TECHNICAL DATA
Duramin-40

Duramin-40
Hardness methods

Vickers

ISO 6507
ASTM E384, E92
JIS B 7725

KnoppISO 4545
ASTM E92
JIS Z 2251

Brinell

ISO 6506
ASTM E10JIS Z 2243

Conversion

Conversions to other hardness methods according to ASTM
E140,
ISO 18265, GB/T 1172

Force range

Test force

Duramin-40 M1/A1/AC1

0.098 – 98,1 N (10gf - 10 kgf)

Duramin-40 M2/A2/AC2

0.098 – 306.5 N (10 gf – 31.25 kgf)

Duramin-40 M3/A3/AC3

0.0098 – 612.9 N (1.0 gf – 62.5 kgf)

Force application

Load cell, closed loop, force feedback system

Test force tolerance< 0.25% for test force above 0.1kgf< 0.5 % for test forces lower than 0.1kgf

Dwell time settings

Standard 10 seconds, user defined up to 250 seconds

TurretMotorized turret

6 position turret, 2 indenter positions, 4 objective positions

Electrical data

Power supply

100 V AC – 240 V AC, 50/60Hz, single phase

Power consumption Max. load Power consumption max.
load

90 W

Idle

67 W

Residual Current Circuit
Breaker (RCCB)
Dimensions

WeightType A, 30 mA is required depending on local regulations.

Width27.4 cm (10.8")

Depth

47.2 cm (18.6")

Height77.5 cm (30.5")

Duramin-40 M1/M2/M3

120 kg (264.6 lbs)

Duramin-40 A1/A2/A3/AC1/AC2/AC3

130 kg (286,6 lbs)

Read method
Overview camera resolution

Automated from Camera Image
Duramin-40 M1/ M2/ M3/A1/A2/A3

No camera

Duramin-40 AC1/AC2/AC3

5 Mpix (optional)

TECHNICAL DATA
Duramin-40

Duramin-40
Overview camera field ofview

Duramin-40 M1/ M2/ M3/A1/A2/A3

No camera

Duramin-40 AC1/AC2/AC3

200 x 160 mm (7.9" x 6.3")Meassurement camera
resolution

5 Mpix or higher

Positions in nosepiece

Duramin-40

6Position in nosepiece foroverview camera

Duramin-40 M1/ M2/ M3/A1/A2/A3

NA

Duramin-40 AC1/AC2/AC3

1

Max no. of Indenters

2

Max No. of Objectives

4

Indenter Shaft

Diameter

3 mm (0.12")

Standard objectives
included

Objective lenses and indenters are ordered separately

Z-AxisMotorized

XY Stage / Anvil

Stage Size

Stage Stroke (travel range)

Duramin-40 M1/M2/M3

Manual XY-stage

Duramin-40 A1/ A2/ A3/ AC1/ AC2/AC3

Automatic XY-stage

Duramin-40 M1/M2/M3

100 x 100 mm (3.9" x 3.9")

Duramin-40 A1/ A2/ A3/ AC1/ AC2/AC3

357 x 208 mm (14" x 8.2")

Duramin-40 M1/M2/M3

25 x 25 mm (0.98" x 0.98")

Duramin-40 A1/ A2/ A3/ AC1/ AC2/AC3

220 x 120 mm (8.7" x 4.7")

Auto Illumination

Yes

Stage Illumination

Yes

Laser/LED Guide

NoSoftware

SystemOperating software

Embedded software for work flow system & tester control

Integrated PC

Yes

Monitor15” touchscreen

Possibility to connect PrinterYes (A4, A3 full color laser printer optional)

Ethernet Connection

Yes

Data Export

5x USB, Ethernet LAN, Wi-Fi, RS232, Bluetooth, HDMI

Data output

XML, CSV,Q-DAS certified (optional)

TECHNICAL DATA
Duramin-40

Duramin-40
Software modules

Duramin-40 M1/M2/M3

Basic Statistics (included)
Data export (included)
Report editor (included)
Test pattern (included)
Editor, CHD (optional)

Duramin-40 A1/ A2/ A3/ AC1/ AC2/AC3

Basic Statistics (included)
Data export (included)
Report editor (included)
Test pattern (included)
Editor, CHD (included)
Kc Fracture (optional)

Sample height

Duramin-40 M1/M2/M3

172 mm (6.8")

Duramin-40 A1/ A2/ A3/ AC1/ AC2/AC3

148 mm (5.8")

Sample weight

25 kg (55 lbs)

Throat depth

170 mm (6.7") (from Center of indenter to back)Safety standards

CE labelled according to EU directives

REACH

For information about REACH. contact your local Struers
officeOperating environment

Surrounding temperature

10-35°C (50-95°F)

Humidity

10%-90% RH non-condensing

Safety Circuit
Categories/Performance
Level

Emergency stop

EN ISO 13849-1 PL c, Category 1

Noise level

A-weighted sound emission pressure level atworkstations< 70 dB(A)Vibration levelDuring operation

Total vibration exposure to upper parts of the body does notexceed 2.5 m/s2.

Stop category 0

Doc. no.: 16607750_A_en
Date of release: 30-06-2023