2450MCTER(E)-v1.0.0

MC Remote Control Software
MCTerminal
Operation Manual

KIMOTO ELECTRIC CO., LTD.

2450MCTER(E)-v1.0.0

2450MCTER(E)-v1.0.0

Abstract
We are delighted that you have decided to use KIMOTO’s analyzers. MCTerminal is a free software that displays
the screen of KIMOTO’s analyzers and data loggers in PC. Also, you can operate the analyzers remotely by themouse operation of PC. Ethernet-HUB supporting 10/100 BASE-T should be necessary to communicate. Microsoft®
Windows-based PC (7, Vista™, XP) is needed to operate the MCTerminal.

Contents
1. INSTALLATION12. NETWORK CONNECTION

2

2.1 Ethernet setting of PC

2

2.2 Connection method between KIMOTO’s analyzer and PC

2

2.2.1 Connection using a cross cable

2

2.2.2 Connection using a Ethernet HUB3

3. OPERATION

4

3.1 Start of MCTerminal

4

3.2 Connection to analyzers

4

3.3 Operation of screen4

3.4 Disconnection

4

4. APPENDIX

5

4.1 Typical IP Address of KIMOTO’s analyzers5

4.2 Version of MCTerminal

5

4.3 Download Site of MCTerminal

5

2450MCTER(E)-v1.0.0

2450MCTER(E)-v1.0.0

1. INSTALLATIONStart the installation process using install program "MCTerminal2.exe". You can download the installer from ourwebsite. (Refer Sec.4.3.)
Wizard screen appears when you open the install program. Install the MCTerminal following the instructions onthe screen.
After installation was performed, you will be able to find MCTerminal to boot on the desktop as below. Also, theMCTerminal can be started from "Start" button -> "Programs" -> "Kimoto Utils" -> "MCTerminal".

-1-2450MCTER(E)-v1.0.0

2. NETWORK CONNECTION
All KIMOTO’s analyzers have an Ethernet port. You can easily download the data to your PC via Ethernet. It may
need some networking knowledge such as administrative privileges of Windows in order to setup the software. If
you have any trouble to setup, please ask to a network administrator or an authority managing the network.

2.1 Ethernet setting of PC

It is necessary to setup Ethernet connection of your PC before data communication. All our analyzers

manufactured after December 2005 have dynamic host configuration protocol (DHCP) that assigns IP addresses
automatically. For other analyzers manufactured before December 2005, it is need to manually setup the IP addressand subnet mask.
For example (manual setting);
IP address: 192. 168. 0. 10
Subnet mask: 255. 255. 255. 0

2.2 Connection method between KIMOTO’s analyzer and PC

There are two methods for communication between our analyzer and PC. One is one-to-one
communication using a cross-cable Another is multiaccess communications using Ethernet HUB.

2.2.1 Connection using a cross cable
The following figure shows a connection using a crossover Ethernet cable. Sometimes the connection cannot beestablished because of specification of the PC. In that case, it is recommended to use HUB as shown in section 2.2.2.

PC

SPM-613KIMOTO

10/100 Base-T Cross Cable
10/100 Base-T Cross Cable
1 RD+2 RD3 TD+
456 TD78-2-1 RD+
2 RD3 TD+4
56 TD782450MCTER(E)-v1.0.0

2.2.2 Connection using a Ethernet HUB
The following figure shows a connection using Ethernet HUB and some straight cables. In the case of theconnection using a HUB, it allows the multiple communications between your PC and our analyzers simultaneously.

12345678UpLinkHUB10/100 Base-T Straight Cable

PCSPM-613

KIMOTO

10/100 Base-T Straight Cable
1 RD+
2 RD3 TD+456 TD78KIMOTO

1 RD+
2 RD3 TD+456 TD78-3-2450MCTER(E)-v1.0.0

3. OPERATION
3.1 Start of MCTerminal"Version" button: displays the version number			of MCTerminal"Connect" button: displays the dialog box			connecting to analyzer"Capture" button: captures screen image of			the analyzer, and saves the			image in BMP format"Exit" button		: ends MCTerminal

3.2

Connection to analyzers(1)Click "Connect" button, and a dialog box
connecting to analyzer is displayed.
Input IP address of the analyzer. Click "Target List"
pull-down menu, and the list of the analyzers thatcan be connected is displayed. Select an analyzer
from the list, then the IP address of the analyzer isinput automatically. Also, IP address can be input
directly.
(2)Click "OK" button. (*)

3.3 Operation of screen

After the connection is established, the screen of

the analyzer is displayed. Click the displayed screenwith the mouse, that can emulate the touch screen
operation of the analyzer.
When you operate the analyzer with MCTerminal;- The actual screen of the analyzer is not displayed.
- The actual touch screen of the analyzer can not sense the touch operation.

3.4 Disconnection

Click "Exit" button to disconnect from the analyzer.

(*) If your OS is Windows 7 or Windows Vista, “Windows Security Alert” would be indicated when you click “OK”
button. In order to allow communication with your analyzer, check-off all boxes and click “Allow Access” button.
Also, administrative privileges are required to allow access.

-4-2450MCTER(E)-v1.0.04. APPENDIX
4.1 Typical IP Address of KIMOTO’s analyzers
Data Logger

192.168.0.110

Nitrogen Oxides Analyzer

192.168.0.120

Ozone Analyzer

192.168.0.130

Sulfur Dioxide Analyzer

192.168.0.140

Particurate Matter Monitor

192.168.0.150

4.2

Version of MCTerminal

4.3

Download Site of MCTerminal

Click "Version" button and you can see the version number of MCTerminal.

The latest installer is located on the web site at;		

http://www.kimoto-electric.co.jp/support/index.html

This manual is applicable for the following versions of software.
PID 2450 Ver.2.0.0 or later-5-

2450MCTER(E)-v1.0.0

2450MCTER(E)-v1.0.0

KIMOTO ELECTRIC CO., LTD.
OFFICE: 3-1 Funahashi-cho Tennoji-ku Osaka 543-0024 JAPANTEL +81-6-6768-3401 FAX +81-6-6764-7040
URL http://www.kimoto-electric.co.jp/