NICHIRYO PRODUCT LINEUPPIPETTE

Autoclavable Multi Channel Pipette

Light & Ergonomic Pipette

MULTI
Light weight and easy to use
even for first-time users.

Multi Channel model of Nichipet EXⅡ
Reduced fatigue for pipetting.

■ The enlarged and stream lined grip fits well with your fingers and can be kept firmly■ Reduced the weight up to 30% ( in-house comparison)■ Easy maintenancein hand and used with much less stress.

■ Reduced ejector load

■ The user-friendly and enlarged push-button can be pressed easily and used for a long

■ Lock mechanism for volume setting

time with less risk of RSI.

■ Durable body

■ The enlarged push button makes it easy to set the volume and there is no need

■ Revolving handle ( 360°) to fixate convenient position in pipetting.

to lock the dial for immediate operation.

Since the volume indicator is made of glass,ultraviolet germicidal of the clean bench will not make it cloud.
From prolonged irradiation, the surface layer of plastic may deteriorate
or discolor, but its performance is not affected.

Specifications
Specifications
Cat. No00-NLE-1000-NLE-2000-NLE-100
00-NLE-200
00-NLE-1000

Volume setting

Digital setting

Volume range(µL)
0.5 – 102 – 20
10 – 100
20 – 200
100 – 1000

Increments (µL)
0.02
0.02
0.2
0.2
2Cat. No
00 -NPM-8V
00 -NPM-8S
00 -NPM-8L
00 -NPM-8K
00 -NPM-12V
00 -NPM-12S
00 -NPM-12L
00 -NPM-12K

Usable tips
BMT2-SSW/SSWR/SSWRB、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB*See p.21-22 for tip selection.

Accuracy and Precision
Cat. No00-NLE-1000-NLE-20

00-NLE-100

17page.
www.nichiryo.co.jp/en/

Volume(µL)
15102
1020
1050
100

ChannelVolume setting

8Digital setting
12

Volume range(µL)
0.5 – 10
5 – 100
20 – 200
30 – 300
0.5 – 10
5 – 100
20 – 200
30 – 300

Increments(µL)
0.01
0.1
0.2
10.01
0.1
0.2
1

Usable tips
BMT2-UT/UTWR/UTWRB、FUT/FUTRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、
FAG/FAGRB
BMT2-K/KR/KRB
BMT2-UT/UTWR/UTWRB、
FUT/FUTRB
BMT2-SG/SGR/SGRB、FSE/FSERBBMT2-SG/SGR/SGRB、
FAG/FAGRB
BMT2-K/KR/KRB*See p.21-22 for tip selection.

Accuracy(%)±6.0*±2.0±1.2±5.0±2.0±1.2±2.0±1.2±1.0

Precision(%)
≦5.0*
≦2.0
≦0.8
≦3.0
≦2.0
≦0.8
≦1.0
≦0.8
≦0.6

Cat. No00-NLE-200

00-NLE-1000

Volume(µL)
20
100
200
1005001000

Accuracy(%)±1.2±1.0±0.8±1.2±1.0±0.7

Precision(%)
≦1.0
≦0.8
≦0.6
≦0.8
≦0.6
≦0.5

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NLE-10 below 1μL depend on the sampling skill and the sampling
condition where pipettes are used.

Accuracy and Precision
Cat. No
00-NPM-8V
00-NPM-12V
00-NPM-8S
00-NPM-12S

Volume(µL)
1
510
1050
100

Accuracy(%)±8.0*±4.0±2.0±3.0*±1.0±0.8

Precision(%)
≦5.0*
≦2.0
≦1.0
≦2.0*
≦0.8
≦0.3

Cat. No
00-NPM-8L
00-NPM-12L
00-NPM-8K
00-NPM-12K

Volume(µL)
2010020030150300

Accuracy(%)±3.0±1.0±0.9±3.0±1.0±0.6Precision(%)
≦0.6
≦0.4
≦0.3
≦1.0
≦0.5
≦0.3

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NPM-8V and 12V below 1μL and NPM-8S and 12S below 10μL depend on the sampling skill and the sampling condition where pipettes are used.

＊visit our web-site for part lists.

page.
www.nichiryo.co.jp/en/18