REPETITIVE SYRINGE DISPENSERFEATURES 27 different volumes of liquid can be dispensed.
 Fast change syringes.
 Light and slender design permits easy handling without fatigue.
 Excellent accuracy and precision.
 Autoclavable syringe.(60.0 mL and 3.0 mL syringes are not autoclavable.)
PARTS AND MATERIALS②

Arrow Mark

①

Fig.1

50mm

Fig.2

1. Thumb Knob (Nylon)
2. Volume Adjustment
Dial(Nylon)
3. Handle Cap(ABS)
4. Handle (ABS)
5. Piston Lever (Aluminium)
6. Syringe Holder(Polypropylene)
7. Piston Holder(Polypropylene)8. Piston Flange(Polypropylene)
9. Syringe Flange(Polypropylene)
10. Syringe (Polypropylene)
11. Piston (Polypropylene &Rubber)

OPERATION
1. Push the piston lever to its lowest position.
2. Select the appropriate syringe. Move the piston of the syringe up and down
several times to ensure smooth operation in the dispenser.
3. Push the syringe flange and the piston flange firmly into the syringe holder andthe piston holder respectively. (Fig.2)
4. Select the volume to be dispensed by turning the volume adjustment dial to theappropriate number. (Fig.1)
5. Place the nozzle of the syringe into the liquid to be dispensed.
6. Pull the piston lever up so that the syringe is filled with liquid. It is recommended
to minimize the volume of air in the syringe to obtain better accuracy andprecision.(Fig.2)
7. Wipe the nozzle of the syringe or tip with a tissue.
8. If the syringe has been fully filled, the thumb knob should be depressed once,
and the liquid expelled before using the dispenser. The dispenser will then deliver
the number of dispenses in accordance with the table attached to the side of thehandle.
9. If the syringe is only partially filled, the dispenser can be used by pressing thethumb knob until it reaches the black line at the end of the syringe.
To ensure that the dispenser maintains its accuracy, do not dispense liquid after
the piston has passed the black line.
10. Place the nozzle or tip of the syringe so that it is touching the inside wall of thereceptacle and depress the thumb knob as far as it will go. The desired volumeof liquid will then be delivered into the receptacle.
11. After finishing dispensing, the liquid remaining in the syringe can be ejected bypushing the piston lever downwards.
ATTACHMENT OF 60.0 ML SYRINGE
1. To mount the adapter or the syringe, tilt theadapter so that square cut on the center hole of
the adapter faces upward, and insert the piston
flange through the hole, keeping the adapter
tilted. Mount adapter on the syringe by turning theadapter clockwise until it will turn no further.
2. Attach the piston flange to the piston holder andthe flange on the adapter to the syringe holder bypushing them against the holders, keeping the flatside of the piston flange faced against the holder.

Fig.3

SPECIFICATIONS

Volume
SG-V2
SG-S
SG-M
SG-N3
SG-L
SG-Y
SG-Z

(0.06mL)(0.6mL)(1.5mL)(3mL)(6mL)(15mL)(60mL)

Accuracy(systematic error)
±2.5%
±1.5%
±1.2%
±1.0%
±0.9%
±0.8%
±1.4%

precision(random error)
≦2.5%
≦1.0%
≦1.1%
≦0.9%
≦0.8%
≦0.8%
≦1.0%

The above specifications were acquired using water at a temperature of 20°C.
The SG-S and SG-M tests were done using syringes having pipette tips on the nozzles.
DISPENSING VOLUMES

0.06mL 0.6mL 1.5mL 3.0mL 6.0mL 15.0mL 60.0mL Maximum
Number(SG-V2) (SG-S) (SG-M) (SG-N3) (SG-L) (SG-Y) (SG-Z) Dispense
11uL
10uL
25uL
50uL 100uL 250uL 1.0mL
58
Volume 22uL
20uL
50uL
100uL 200uL 500uL 2.0mL
28Adjust33uL
30uL
75uL
150uL 300uL 750uL 3.0mL
18ment44uL
40uL 100uL 200uL 400uL 1000uL 4.0mL
13Dial
55uL
50uL 125uL 250uL 500uL 1250uL 5.0mL
10Size ofSyringe

ORDERING INFORMATION

Cat. No.
00-8100
00-SG-V200-SG-S
00-SG-M
00-SG-N300-SG-L
00-SG-Y00-SG-Z
00-SG-ZA
00-8100S

Description
One instrument with one each of 6.0mL, 1.5mL and
0.6mL syringe
0.06mL syringe without tip
0.6mL syringe with tip
1.5mL syringe with tip
3.0mL syringe without tip
6.0mL syringe without tip
15.0mL syringe without tip
60.0mL syringe without tip, but with one adapter tomount the syringe to the handle
Adapter for 60mL syringeStand to hold two handles with syringesQuantity
1100
100
100
100
100
5050
11

For repair service or information you may contact your local distributor.

MANUFACTURER:
SB-81002E