NICHIRYO PRODUCT LINEUPPIPETTE

Fixed Volume & Three Volume Micro Pipette

Variable Volume Macro Pipette

MODEL 3100
■The light and slender design along with the smooth movement of the plunger reduces the operatorʼs fatigue.

■ Four different volume models from 1 mL to 10mL

■ Built-in tip ejector■ Light weight and slender body

■ PTFE ( Fluorine resin ) is used in the airtight chamber of the instrument,■ Easy to operate

so the instrument keeps its air tight seal and precise reproducibility for long hours.

■ Liquid touches only disposable tip■ PTFE piston ensures complete sealing of the syringe.

Nichipet FⅡ▷Suitable when only one volume is needed.
▷A wide range of standard volumes from 10µL – 1000µL.
▷Other volumes within the range available for special order.
Nichipet VⅡ▷Three different volume models▷Volume change by simply turning the knob.
Specifications
TypeFⅡVⅡCat. No
00-NPF2-1000-NPF2-2000-NPF2-2500-NPF2-3000-NPF2-5000-NPF2-10000-NPF2-20000-NPF2-50000-NPF2-100000-NPV2-S00-NPV2-M00-NPV2-L

Volume setting

FixedThree variablesettings

Volume range(µL)
1020253050100200500
1000
10/20/50
50/100/200
200/500/1000

Accuracy(%)
±1.2
±0.8
±0.8
±0.8
±0.6
±0.6
±0.6
±0.5
±0.5
±1.2 – ±0.7
±0.7 – ±0.5
±0.7 – ±0.5

Precision(%)
≦0.4
≦0.3
≦0.3
≦0.3
≦0.3
≦0.3
≦0.2
≦0.2
≦0.2
≦0.6 – ≦0.3
≦0.4 – ≦0.3
≦0.3 – ≦0.2

Usable tips
BMT2-SSW/SSWR/SSWRB、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB

Specifications

BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB

Cat. No00-31N-1000
00-31N-2000
00-31N-5000
00-31N-10000● Accuracy and Precision● Disposable tips
Volume range(mL)
0.2 – 1.0
0.4 – 2.0
1.0 – 5.0
2.0 – 10.0

Increment(mL)
0.1
0.1
0.5
0.5

Cat. No
00-MPT-100-MPT-200-BMT2-X
00-BMT2-Z

Volume range(mL)
0.2 – 1.0
0.4 – 2.0
1.0 – 5.0
2.0 – 10.0Color
Clear
Clear
Clear
Clear

Q'ty(bag)
1000
200
200200± 1.0% / ≦ 1.0%

*See p.21-22 for tip selection.

Glass Tip Pipette

Three Volume Ultra Macro Pipette

Nichipet ECO

MODEL 5100®

■The accuracy and precision of a micropipette with the stability and reliability■Two models, each with three standard volume settings.

of glass tips.

■ Volume change done by placing a volume spacer in the pipette.

■The glass tip enables bias free liquid handling for endocrine disruptor's■ Any special volume spacer between 30 to 50 mL for 5100L is availableanalysis.

for special order.

■ Four different sizes covering a wide volume range. ( 20µL – 10000μL )■ Glass tip is secured to the unit with a double O-ring design.
■ Comes with a PTFE nozzle and ceramic piston.

Glass Tip

Specifications
Cat. No
00-NPC-200
00-NPC-1000
00-NPC-5000
00-NPC-10ML● Glass tips
Volume range(µL)
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Increment(µL)
0.2
1.0
10.0
50.0

Accuracy(%)±6.0 – ±3.0
±5.0 – ±3.0±5.0 – ±3.0
±6.0 – ±3.0

Precision(%)≦6.0 – ≦3.0≦5.0 – ≦3.0
≦5.0 – ≦3.0
≦6.0 – ≦3.0

Usable tips
Glass Tip S
Glass Tip L
Glass Tip X
Glass Tip Z

Cat. No00-GLT-S00-GLT-L
00-GLT-X
00-GLT-Z

Volume range(µL)
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Q'ty(box)
10
10
1010

Further research into "Environmental Hormone" is strongly requested to clarify the precise cause and effect on the human system and wild living
things. Glass tips rather than standard plastic pipette tips are more suitable for this research.
Soil

Natural Water

PlantsSAMPLING
EXTRACATION
1st CONCENTRATIONPreparation on crystal filter urethane form
Sample transfer with Nichipet ECO
By rotary evaporator

REFINEMENT

Clean up with Chromatograph tube.
The sample is transferred with Nichipet ECO

Specifications

2nd CONCENTRATION

By concentrator.
The sample is transferred with Nichipet ECO

Cat. No00-5100L

FINAL ANALYSIS

19Foods

page.
www.nichiryo.co.jp/en/

By Gas Chromatography● Accuracy and Precision

30.0

Volume setting(mL)
40.0

50.0

● Disposable tips
Cat. No00-51TL

Volume range(mL)
50.0

Q'ty(bag)5± 1.0% / ≦ 1.0%

＊visit our web-site for part lists.

page.
www.nichiryo.co.jp/en/

20