RION Portable Multi-function Measuring System

The groundbreaking multi function measuring system from RION
Compact design, easy and intuitive operation
Wireless connections
Use it anytime anywhere!

Analysis result display examples
FFT analysis
RIONOTE enables you to perform FFT analysis on multiple channels simultaneously.
The results are shown in clear graphs on the large color screen, in real time, or fromstored data when using the recall function. A marker allows you to scroll through thedata, and enables the readout of the level of a frequency of interest.

is combining the newest tquality, ease of use and economical sense.
which can be configured to up to 16 channanywhere wireless. The Main Control Unit isprogram of your choice. All on a large colo
both programs and hardware for this mea

Transfer function
The transfer function represents the relation between an input signal and output
signal in the frequency domain, allowing the determination of amplitude and phase.
In this mathematical calculation category, the RIONOTE supports coherence function
and cross spectrum processing.

Waveform recording
By using the waveform recording program, it is possible to display and record the time
waveform of the incoming signal(s). Available recording time depends on the numberof input channels and the selected frequency range. The figure below shows a time
waveform displayed on the screen of the Main Control Unit.

Main Control Unit and Amplifier
Supports direct connection of microphones andpiezoelectric accelerometers.

Sensor amplifier slides into theunderside of the Main Control UnitRIONOTE System Configuration
Waveform post processing
After completing waveform recording (as explained above), the stored waveforms
can be displayed on the Main Control Unit's large screen, and played back by using
the earphone jack output. Moreover, various secondary post processing functions
for the waveform data are available in the Main Control Unit, including FFT analysis
as shown in the screen example below.

BNC-BNC Coaxial Cable
EC-90 Series
1/2 inchelectret condenser
microphone
UC-52/57/59

Preamplifier NH-22
BNC Adapter
VP-52C

Accelerometer
PV-91C/91CH

Accelerometer cable
VP-51 SeriesWith ampliﬁer

Tacho pulse inpu
General purpose

Voice memo
input connector

Charge Converter
VP-40
Monitor output
connector

Accelerometer Cable
Accelerometer
PV-08A/85/90H etc. VP-51 series
Accelerometer Cable
VP-51 series

I/O
BNC Adapter
VP-52C

Accelerometer
Charge converter
PV-08A/85/90H etc. VP-42

Lithium-ion
Rechargeable
Battery
BP-30 (spare)＊Connection cable requirements differ.

VibrationmeterSound level
meter

Rotation, distortion,temperature, etc.

＊Up to 4 units cantechnology with the traditional virtues of RION;. RIONOTE consists of a Main Control Unit SA-A1
nels and allowing you to perform measurementss easy and intuitive to operate, with the dedicatedor touch screen. RION will continuously develop
asuring system of the future.

Octave band analysis
Real time analysis of noise or vibration levels for evaluation anddesigning countermeasures is usually performed by means of octave
band analysis (using either octave bands or 1/3 octave bands). The
below screen sample of the RIONOTE displays octave analysis resultsin 4 channels as a graph and numeric values at the same time.

RIONOTE intuitive user interface
Lets the user select the required program for the respective purpose:
SX-A1FT (FFT analysis), SX-A1RT (octave band analysis), orSX-A1WR (waveform recording). The right side of the screen provides
access to various settings.

RIONOTE also enables the use of a wireless dock or wireless sensor
amplifiers to avoid the cost and hassle of cables. A plurality of wirelessdocks and wireless sensor amplifiers can be used simultaneously, up to
16 channels, to store the measured data in the Main Control Unit as wellas in the memory of wireless dock or wireless sensor amplifiers.

Wireless Dock (and Amplifier)Separate type wireless dock and amplifier(2 channel or 4 channel configuration)＊Selling of Wireless dock
(SA-A1WD) differs fromeach country.
Please contact us forfurther questions.
Analysis program

4 channel Amplifier
SA-A1B4(Input 4 channel ＋Rotation pulse input)

ut /e DC input

2 channel Amplifier
SA-A1B2(Input 2 channel ＋Rotation pulse input)
AC adapter
NE-20P(Supplied)

Octave band analysis

LAN
USB (mini B)
USB (Type A)
FFT analysis

SD card（Max. 32 GB）

SA-A1

Waveform recording

Waveform Analysis Software

Wireless Dock
Amplifier＋SA-A1WD
SA-A1B4/B2

n be used simultaneously

AS-70

RIONOTE calibration screenServes for calibration of microphones or accelerometers connectedto the SA-A1.

SX-A1FT, RIONOTE Program for FFT Analysis■ Ordering Information
Product name

General real-time analysis processing

Product number

RIONOTE 2 channel FFT Analyzer

SA-A1FTB2

Processing outline

FFT analysis (non-continuous frames when used in real time)

RIONOTE 4 channel FFT Analyzer

SA-A1FTB4

Number of channels

Max. 4 channels

RIONOTE 2 channel Octave Analyzer

SA-A1RTB2

Trigger Trigger modes

RIONOTE 4 channel Octave Analyzer

SA-A1RTB4Trigger source

RIONOTE 2 channel Frequency Analyzer (FFT and Octave)

SA-A1FTRTB2

Trigger position

RIONOTE 4 channel Frequency Analyzer (FFT and Octave)

SA-A1FTRTB4RIONOTE Program for FFT Analysis

SX-A1FTRIONOTE Program for 1/3 Octave Analysis

SX-A1RT

Arithmetic functions

Waveform, External, Rotation speed
Nー (N: number of analysis points)±2

Time domain waveform for 1 frame, Power spectrum, Cross spectrum,
Transfer function, Coherence

Options
Product name

Free, Single, Repeat

Product number

Window functions

Rectangular, Hanning, Flat-top, Exponential, Force

Analysis frequencies

20 kHz, 10 kHz, 5 kHz, 2 kHz, 1 kHz, 500 Hz, 200 Hz, 100 Hz

Number of analysis points

256, 512, 1 024, 2 048, 4 096, 8 192, 16 384

Averaging and otherLinear, Exponential, Max HoldSA-A1WDprocessing functions

Lithium-ion Rechargeable Battery (spare)

BP-30

Number of averaging runs 1 to 1 024

32 GB SD Card

MC-32SD3

General post-analysis processing

2 GB SD Card

MC-20SD2

Outline

Voice Memo Microphone

BSHSM03BK

Number of channels

Max. 4 channels

Monitor Earphone

ATH-C320-BK

Arithmetic functions

Time waveform for 1 frame, Power spectrum, Cross spectrum,

Wireless Dock

FFT analysis of WAVE files recorded with WR function

Transfer function, Coherence, Partial overall

Shoulder Belt

VA-12015

LCD Protector-Window functions

Rectangular, Hanning, Flat-top, Exponential, ForceCCLD 4 mA Modification (factory option)-Number of analysis points

1 024, 2 048, 4 096, 8 192, 16 384, 32 768

Overlap ratio

0 %, 25 %, 50 %, 75 %■ Specifications
RIONOTE Main Control Unit SA-A1, RIONOTE 4 channel / 2 channel Amplifier SA-A1B4/B2

Averaging and otherLinear, Exponential, Max Hold

Input section

Number of averaging runs

Number of channels

processing functions
1 to 1 024

4 (2), BNC connectors

Max. input voltage

±13 V

CCLD

2 mA

SX-A1RT, RIONOTE Program for 1/3 Octave Analysis
Standard compliance

24 V (4 mA Factory option)

JIS C1513 Class 1, JIS C1514 Class1,IEC 61260:1995 Class1, ANSI S1.11-2004 Class1

Amplifier section

Band filter center frequencies and number of bands

Frequency Range

DC to 20 kHz or 0.25 Hz to 20 kHz

Input range-40 dB to 20 dB, 20-dB steps, 0 dB ref. Vrms = 1 V

Octave bands

0.5 to 16 000 Hz, 16 bands Max. 4 channels

Residual noise

At range full-scale: -85 dB or less (0 dB range, AP level)

1/3 octave bands

0.4 to 20 000 Hz, 48 bands Max. 3 channels

Dynamic range

100 dB or better (0 dB range, fs = 51.2 kHz, 400 line FFT noise level)

Instantaneous value data

Time weighted level Lp, Time averaged level Leq, Time weighted

Phase difference±1 deg. or less (1 Hz to 20 kHz, same input range)(every 100 ms)

maximum level Lmax

Processing value data

between channels

Time averaged level Leq, Sound exposure level LE,
Time weighted maximum level Lmax, Time weighted minimum level Lmin,

A/D converter section

Time percentile level LN (5, 10, 50, 90, 95, 33.3), max. 5 values

A/D converter

24 bit, delta-sigma type, simultaneous sampling

Sampling frequencies

51.2 kHz, 25.6 kHz, 12.8 kHz, 5.12 kHz, 2.56 kHz,1.28 kHz, 512 Hz, 256 Hz

Store function

Auto/Manual

10.1 inch TFT color LCD, 1 280 x 800 pixels, transmissive typeTime weighting

F (Fast) 125 ms, 630 ms, S (Slow) 1 s, 10 s

Multi-touch (2 points), projected capacitive typecharacteristics

Display
Touch panel

Frequency weightingInput/output section
USB A x 1, mini B x 1

characteristics

Earphone jackYes Stereo mini jack, φ3.5Trigger

SD card slot

Yes (SDHC support, max. 32 GB)

USB

A, C, Z

Trigger modes

Free, Single, Repeat

Trigger source

AP level, Band level, External signal, Time

Tacho pulse input

SX-A1WR, RIONOTE Program for Waveform recording (Installed in SA-A1 main unit)

CommonNumber of channels

1, BNC connector

Number of recording

Input voltage range

0 to 12 Vchannels
Frequency range

Tacho
Measurement rotation 5 000 pulse/s

Quantization

speed range

TriggerTrigger modes
Trigger source

General purpose
A/D converter

10 bit successive approximation type

Sampling frequency

Voice memo marker function

1 to 4 channels + rotation or General purpose DC
20 kHz, 10 kHz, 5 kHz, 1 kHz, 500 Hz, 100 Hz
16 bit/24 bit
Free, Single, Repeat
Waveform, Time, External, Rotation speed
Yes

Approx. 10 Hz

Monitor output (playback)Allows listening to recorded data (51.2 kHz, 25.6 kHz, 12.8 kHz only)

External trigger

Open collector supported, internal pull-up 3.3 V

Recorded data

WAVE format

Power supply

Li-Ion battery (battery life approx. 4 hours, depending on usage conditions), AC adapter

Dimensions, Weight

Precautions regarding waterproofing
Before use, verify that the connector cover on the side of the unit is firmly closed.
To maintain the water-resistant rating, the internal packing of the enclosure must
be replaced every two years (at cost).

40 (H) x 275 (W) x 188 (D) mm
SA-A1: 1 200 g (incl. 280 g battery, SA-A1B4 mounted)

Water-resistant rating

Equivalent to IP54

Operating temperature range -10 °C to +50 °C using AC adapter, max. 90 % RH (no condensation)
Supplied accessories

Rechargeable Li-Ion battery, BP-30 x 1, AC adapter NE-20P x 1

RIONOTE Wireless Dock, SA-A1WD (and Amplifier SA-A1B4/B2)
Input

4 or 2 channels (Amplifier SA-A1B4/B2 needed)

Signal transfer to LAN port

Ethernet 100 base-TX

main platform

Wireless

Distance of wireless transfer

WLAN (IEEE802.11a/b/g/n, 2.4/5 GHz)
about 50 m＊Memory

SD card (SDHC support, max 32 GB)Power supply

8 IEC R6 (sizeAA) batteries(alkaline or nickel-hydride), AC adapter

Dimensions, WeightApprox. 42 (H) × 193 (W) × 95 (D) mm, Approx. 500 g (incl. battery)

Water-resistant rating

IP grade IP54 equivalent (same as main unit)＊ Depending on usage conditions

RION Co., Ltd. is recognized by the JCSS which uses ISO/IEC 17025 (JIS Q 17025) as an accreditation standard and bases its
accreditation scheme on ISO/IEC 17011. JCSS is operated by the accreditation body (IA Japan) which is a signatory to the Asia
Pacific Laboratory Accreditation Cooperation (APLAC) as well as the International Laboratory Accreditation Cooperation (ILAC).
The Quality Assurance Section of RION Co., Ltd. is an international MRA compliant JCSS operator with the accreditation number
JCSS 0197.
＊ Windows is a trademark of Microsoft Corporation.

ISO 14001 RION CO., LTD.
ISO 9 0 0 1 RION CO., LTD.

＊ Specifications subject to change without notice.

Distributed by:https://rion-sv.com/
3-20-41, Higashimotomachi, Kokubunji, Tokyo 185-8533, Japan
Tel: +81-42-359-7888 Fax: +81-42-359-7442
This product is environment-friendly. It does not include toxic chemicals on our policy.
This product is certified to an International Protection rating of IP54 (dust protected and resistant to splashing water).
This leaflet is printed with environmentally friendly UV ink.
1405-81904.P.D