NANOSECOND LASERS

ANL seriesHigh Energy and High Repetition
Rate DPSS Nanosecond Lasers

Highly Customizable
to Meet CustomerNeeds
F E AT U R E S▶ Up to 1 J at 1064 nm output

pulse energy▶ Up to 1 kHz repetition rate

▶ 2 – 4 ns or 5 ns pulse duration▶ Spatial flat top beam profile▶ Low maintenance costs▶ Various customizing possibilities totailor for specific applications▶ Optional second and third

harmonics generators▶ High efficiency diode pumpingchambers▶ 1×2 m laser head footprint▶ Internal system diagnostics▶ Optional industrial grade, portable

ANL series electro-opticallyQ-switched nanosecond Nd:YAGlasers deliver high energy pulses athigh repetition rates.
A diode-pumped Q-switchednanosecond laser, based on industrytested technology is used as a master
oscillator of the system. It produces
high-intensity, high-brightness
pulses and is well suited for further
amplification in linear amplifiers forhigh-energy flat-top output pulses.
Employing electro-optical cavity
dumping, the master oscillator canproduce pulses which are as short asseveral ns with uniform beam profileand low divergence.
Power amplifiers are a chain of
low-maintenance diode-pumped
single and double pass amplifiers

R E V. 20200123

where pulses are amplified up to therequired energy. During amplification,
spatial beam shaping is employed inorder to get a flat top shape at theoutput. Optional second and third
harmonic generators are based onangle-tuned nonlinear crystals placedin heaters.
For convenience, PC software forWindows™ (LabVIEW™ drivers aresupplied as well) is used for laser
operation, monitoring and internalsystem diagnostics.
To tailor the laser for specificapplications or requirements, various
customization possibilities areavailable such as industrial grade,portable laser housing with integrated
power supplies and cooling units.

laser housing with integrated
power supplies and cooling units

S AVA N O R I U AV. 237, LT-02300 V I L N I U S, L I T H UA N I A T E L +370 5 2649629 E-M A I L S A L E S@E K S P L A.CO M W W W.E K S P L A.CO M

25

ANL series

NANOSECOND LASERSS P E C I F I C AT I O N S ¹⁾
Model

ANL 2001k

ANL 4001k

ANL 1k200> 200 mJ

> 400 mJ

> 1000 mJ

MAIN SPECIFICATIONS
Pulse energy
at 1064 nm
at 532 nm ²⁾–Pulse energy stability (StdDev): ³⁾
at 1064 nm

1.5 %at 532 nm–Power drift ⁴⁾±2%

Pulse duration ⁵⁾

2 – 4 ns

~ 5 nsRepetition rate

1000 Hz200 Hz

Polarization at 1064 nm

horizontal

Optical pulse jitter ⁶⁾–Linewidth–

Beam profile

Hat-Top (at laser output), without diffraction rings

Typical beam diameter ⁷⁾
Beam divergence ⁸⁾
Beam pointing stability~6 mm~10 mm< 1.0 mrad

< 0.5 mrad
± 30 µrad ³⁾

PHYSICAL CHARACTERISTICS
Laser head (W × L × H)

1000 × 2000 × 490 mm

Power supply unit (W × L × H)

553 × 600 × 700 mm

Umbilical length

2.5 m

OPERATING REQUIREMENTS
Facility water consumption (max 20° C)

10 l/minAmbient temperature
Relative humidity

208, 380 or 400 V AC, three phase, 50/60 Hz

Power consumption

2)

3)

4)

5)

6)

7)

8)
9)

10 l/min20 – 80 % (non-condensing)Power requirements ⁹⁾

1)

14 l/min
22 ± 2 °C<10 kW<12 kW

Due to continuous improvement, all specifications
subject to change without notice. Parameters marked
typical may vary with each unit we manufacture.Unlessstated otherwise, all specifications are measured at1064 nm and for basic system without options.

<6 kW

DANGERVISIBLE AND/OR INVISIBLE LASER RADIATIONAVOID EYE OR SKIN EXPOSURE TO DIRECT
REFLECTED OR SCATTERED RADIATIONFor NL94X-SH harmonic generator option. Harmonic
outputs are not simultaneous; only single wavelength
beam is present at the output at once.

Nd:YAG 1064 nm, 532 nm
Max. 4 J, pulse 5 ns
CLASS IV LASER PRODUCTStandard deviation value averaged over 30 s after
20 minutes of warm-up.

Deviation from average value measured over 8 hoursof operation when room temperature variation is lessthan ±2 °C.
Measured with photodiode with 100 ps rise time andoscilloscope with 600 MHz bandwidth.
Standard deviation value, measured with respect totriggering pulse.

Beam diameter is measured at 1064 nm at laser
output at the 1/e² level and can vary with each unit wemanufacture.

Typical beam profile of ANL4001k laser

Full angle measured at the 1/e² level at 1064 nm.

Mains voltage should be specified when ordering.

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067 FAX:048-871-0068

e-mail:voc@phototechnica.co.jp

26

S AVA N O R I U AV. 237, LT-02300 V I L N I U S, L I T H UA N I A T E L +370 5 2649629 E-M A I L S A L E S@E K S P L A.CO M W W W.E K S P L A.CO M   R E V. 20200123

Femtosecond Lasers

PICOSECOND TUNABLE SYSTEMS
PGx01 ▪ PGx03 ▪ PGx11 ▪ PT277

Picosecond Lasers

PGx01 seriesHigh Energy
Broadly TunableOPA
F E AT U R E S▶ Ultra-wide spectral range

from 193 to 16 000 nm

Picosecond
Picosecond Tunable SystemsSystems▶ High peak power (>50 MW)

ideal for non-linear spectroscopy
applications▶ Narrow linewidth <6 cm⁻¹

(for UV < 9 cm-1)▶ Motorized hands-free tuning

in 193 – 2 300 nm or
2 300 – 16 000 nm range▶ PC control via USB port (RS232 isoptional) and LabVIEW™ drivers

Nanosecond Lasers

Travelling Wave Optical Parametric
Generators (TWOPG) are an excellent
choice for researchers who need
an ultra‑fast tunable coherent lightsource from UV to mid IR.

DesignNanosecond Tunable Lasers
High Intensity Lasers
Other Ekspla Products

32

The units can be divided into several
functional modules:▶ optical parametric generator (OPG);▶ diffraction grating based linewidth

narrowing system (LNS);▶ optical parametric amplifier (OPA);▶ electronic control unit.

The purpose of the OPG module isto generate parametric superfluorescence (PS). Spectral properties of thePS are determined by the propertiesof a nonlinear crystal and usually varywith the generated wavelength.
In order to produce narrowband
radiation, the output from OPG isnarrowed by LNS down to 6 cm⁻¹and then used to seed OPA.
Output wavelength tuning isachieved by changing the angle ofthe nonlinear crystal(s) and grating.
To ensure exceptional wavelength
reproducibility, computerized control
unit driven precise stepper motors
rotate the nonlinear crystals anddiffraction grating. Nonlinear crystal
temperature stabilization ensureslong‑term stability of the output
radiation wavelength.
In order to protect nonlinear crystals
from damage, the pump pulse
energy is monitored by built-in
photodetectors, and the control unit
produces an alert signal when pumppulse energy exceeds the preset value.
For customer convenience thelaser can be operated from masterdevice or personal computer throughUSB (VCP, ASCII commands), RS232(ASCII commands) or LAN (REST API)
interfaces or from remote control padwith backlit display that is easy toread even while wearing laser safety
glasses.

▶ Remote control via keypad

A P P L I C AT I O N S▶ Nonlinear spectroscopy:

vibrational-SFG, surface-SH,Z-scan▶ Pump-probe experiments▶ Laser-induced fluorescence (LIF)▶ Other laser spectroscopyapplications

Available models
ModelFeatures

PG401

Model has a tuning range from 420 to 2300 nm and is
optimized for providing highest pulse energy in the visible partof the spectrum. The wide tuning range makes PG401 units
suitable for many spectroscopy application.

PG501-DFG

Model has a tuning range from 2300 to 16000 nm.
The PG501-DFG1 model is the optimal choice forvibrational-SFG spectroscopy setups.

SAVANORIU AV. 237, LT-02300 VILNIUS, LITHUANIA TEL +370 5 2649629 E-MAIL SALES@EKSPL A.COM WWW.EKSPL A.COM   REV. 20210413

S P E C I F I C AT I O N S 1)
PG401-SH

DUV–SH–Signal

420 – 680 nm

Idler

PG401-DUV

Output pulse energy
Linewidth> 100 µJ at
300 nm< 6 cm-1

Max pulse repetition rate
Scanning step
Signal––< 9 cm-1> 50 µJ at
200 nm~4 mm

Beam polarization–

Beam divergence 4)
Signal< 2 mradhorizontal

Idler

horizontal

Typical pulse duration

2300 – 16000 nm> 250 µJ at 3700 nm,> 40 µJ at 10000 nm> 250 µJ at 3700 nm,> 80 µJ at 10000 nm< 6 cm-1–1 nm

Typical beam size 3)2300 – 10000 nm

50 Hz

0.1 nmIdler––> 1000 µJ at450 nm2)

PG501-DFG2

193 – 209.95 nm

210 – 340,
370 – 419 nm

740 – 2300 nm

DFG

PG501-DFG1–~3 mm~15 ps~12 ps

–

10 mJ

~9 mm
–

verticalhorizontal––

Picosecond Lasers

PG401

Tuning range

Picosecond
Picosecond Tunable Systems
Systems

Model

Femtosecond Lasers

PGx01 series

PICOSECOND TUNABLE SYSTEMS~20 psat 1064 nm
5)Beam divergence
Beam profile

Size (W x L x H)

456 × 633 ×244 mm

OPERATING REQUIREMENTS

PL2231-50-TH,
PL2251A-TH

15 mJ

PL2231A-50-SH,
PL2251B-SH

Due to continuous improvement, allspecifications are subject to change withoutnotice. Parameters marked typical are notspecifications. They are indications of typicalperformance and will vary with each unit wemanufacture. Unless stated otherwise, allspecifications are measured at 450 nm for
PG401 units, 3000 nm for PG501 units and300 nm for PG401SH units and for basic system
without options.

456 × 1031 × 249 ± 3 mm

100 – 240 V AC single phase, 47 – 63 HzPower consumption< 100 W3)
4)
5)

6)

Beam diameter is measured at the 1/e² level.
Full angle measured at the FWHM point.

If a pump laser other than PL2250 or PL2230
is used, measured beam profile data should bepresented when ordering.

Should be specified if non-EKSPLA pump laser
is used.

See tuning curves for typical pulse energies
at other wavelengths. Higher energies areavailable, please contact Ekspla for moredetails.

REV. 20210413

10 mJ

15 – 30 °C

Power requirements2)< 0.5 mrad

6 mJ

30 ± 5 ps

Room temperature1)

PL2231-50-TH,
PL2251A-TH

2 mJ

homogeneous, without hot spots, Gaussian fit >90 %Pulse duration 6)
PHYSICAL CHARACTERISTICS–Nanosecond Tunable Lasers

Recommended pump source–SAVANORIU AV. 237, LT-02300 VILNIUS, LITHUANIA TEL +370 5 2649629 E-MAIL SALES@EKSPL A.COM WWW.EKSPL A.COM

DANGERVISIBLE AND/OR INVISIBLE LASER RADIATIONAVOID EYE OR SKIN EXPOSURE TO DIRECTREFLECTED OR SCATTERED RADIATION
Tunable, 210 – 16 000 nm
Max. 1 mJ, pulse 20 ps
CLASS IV LASER PRODUCTHigh Intensity Lasers

at 532 nm–Other Ekspla Products

at 355 nm

Nanosecond Lasers

PUMP LASER REQUIREMENTS

Pump energy

33

Femtosecond Lasers

PGx01 series

PICOSECOND TUNABLE SYSTEMSC U S TO M I Z E D F O R S P E C I F I C R E Q U I R E M E N T S
Please note that these products arecustom solutions tailored for specificapplications or specific requirements.

Picosecond Lasers

Interested? Tell us more about yourneeds and we will be happy toprovide you with tailored solution.

PG401-DFG1 provides:

PG402 features:▶ The broadest hands-free tuning range –▶ Gap-free tuning range▶ It can be further extended up to 16000▶ Linewidth < 18 cm-1

from 420 to 10000 nm

410 – 709, 710 – 2300 nm

nm with -DFG2 option. It should benoted, that for the 8000 – 16000 nm
range a different nonlinear crystal isused, and exchange of the crystals needs
to be done manually

TUNING CURVES
Wavenumber, 103 × cm-1
8.3
6.2
5.0

52.1

Wavenumber, 103 × cm-1
51.0
50.0
49.0

45.4

48.0

38.4

Wavenumber, 103 × cm-1
33.3
26.3
29.4

140

PG401100

23.8PG401-SH

PG401-DUV
1200.8Pulse energy, µJPulse energy, mJ

1.0

4.1

0.60.480

Pulse energy, µJ

12.5604010080
60400.2200.0
4000192

800

1200
1600
Wavelength, nm

2000

2400

Pump energy: 10 mJ at 355 nm

0200
204
Wavelength, nm

196208

Fig 2. Typical PG401-DUV model tuningcurve

260300340
Wavelength, nm

380420

Wavenumber, cm-1
1000.0833.3
714.2

120625.0
–DFG2

100

Pulse energy, µJ

2502001501005080604020Air absorption

02000 3000 4000 5000 6000 7000 8000 9000 10000
Wavelength, nmFig 4. Typical PG501-DFG1 tuning curvein 2300 – 10000 nm range

080001000016000

Fig 5. Typical PG501-DFG2 tuning curvein 8000 – 16000 nm range

Pump energy: 7 mJ at 1064 nm

Pump energy: 15 mJ at 1064 nm

110317

Hx00

Other Ekspla Products

PL223x or PL225x

1031
1170

1200

456

PGx01

100

High Intensity Lasers

LRECOMMENDED UNITS ARRANGEMENTO N O P T I C A L TA B L E

12000
14000
Wavelength, nm

PL223x or PL225x

PGx01

456

Nanosecond Tunable Lasers

Note: The energy tuning curves
are affected by air absorption duenarrow linewidth. These pictures
present pulse energies where airabsorption is negligible.

1250.0–DFG1 or DFG2 option

300220

Fig 3. Typical PG401-SH model tuningcurve. Pump energy: 10 mJ at 355 nm

Wavenumber, 103 × cm-1
5.00 3.33 2.50 2.00 1.66 1.43 1.25 1.11 1.00
350

Pulse energy, µJ

Nanosecond Lasers

Fig 1. Typical PG401 model tuning curve

20600
456

Picosecond
Picosecond Tunable Systems
Systems

34

25.0LFig 6. Arrangement of pump laser and PGx01 unit on optical table

1031

1500

Fig 7. Recommended arrangement of pump laser andPGx01-DFGx unit on optical table

SAVANORIU AV. 237, LT-02300 VILNIUS, LITHUANIA TEL +370 5 2649629 E-MAIL SALES@EKSPL A.COM WWW.EKSPL A.COM   REV. 20210413

O U T L I N E D R AW I N G S

456

272*

235
c

170 ±5LFig 8. PG401 external dimensions

OUTPUTS PORTS
L, mma, mmb, mmc, mm

PG401-SH

838

380××380×250

50110

Port 1Port 2

210 – 340 nm, 370 – 419.9 nm,420 – 680 nm, 740 – 2300 nm–420 – 680 nm, 740 – 2300 nm–210 – 340 nm, 370 – 419 nm,420 – 680 nm, 740 – 2300 nm

192 – 209.95 nm

90

PG501

Output ports
9550

For SFG optional 532 nm
output port 2.
Input ports

Nanosecond Tunable Lasers

456901031

Fig 9. PG501 external dimensions

O R D E R I N G I N F O R M AT I O N
PG401-DUV
Model
PG4xx → 355 nm pump
01 → travelling wave,
narrowed linewidth
02 → travelling wave,
not narrowed
11 → synchronous pumping,narrowed

PG501-DFG1

Optional tuning range extension
DUV→ 193 – 209.95 nm
SH→ 210-340 nm &370-420 nm
Custom products, tailored forspecific applications. Inquire forother specifications.

DFG1
DFG2→ 2300 – 10000 nm;>250 μJ at 3700 nm→ 2300 – 16000 nm

Note: Laser must be connected to the mains
electricity all the time. If there will be no mains
electricity for longer that 1 hour then laser (system)needs warm up for a few hours before switching on.

Model
PG5xx → 532 nm pump
01 → travelling wave,
narrowed linewidth
Tuning range
DFG1 → 2300 – 10000 nm;>250 μJ at 3700 nm
DFG2 → 2300 – 16000 nm

www.phototechnica.co.jp

High Intensity Lasers1026

×Other Ekspla Products

PG401-SH/DUV

380

239...249

633

165...175

PG401

Nanosecond Lasersb

Picosecond
Picosecond Tunable Systems
Systems

Picosecond Lasers

Input panel
aModelFemtosecond Lasers

PGx01 series

PICOSECOND TUNABLE SYSTEMS〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067 FAX:048-871-0068

e-mail:voc@phototechnica.co.jp
REV. 20210413

SAVANORIU AV. 237, LT-02300 VILNIUS, LITHUANIA TEL +370 5 2649629 E-MAIL SALES@EKSPL A.COM WWW.EKSPL A.COM35