Particulate Matter Monitor
model PM-711
PM-711 is a continuous monitor that measuressuspended particulate matter (SPM) in ambient air
by a beta ray attenuation method complying withJIS B 7954.

High accuracy monitoring using PTFE filter has avery small effect for gas components or moisture.

Large capacity memory to save measured data forover a year.

Simple operation on a large color LCD touch paneloffers a lot of information.

USB memory for general-purpose is available todata extraction.

Easy to replacement of filter roll and maintenanceby adopting a fold type slide front panel.

Ethernet interface is available as standard equipment for digital telemetering.

General Specifications
Measuring object

SPM in the ambient air

Stability of flow rate

±5%/10days of setting flow rate

Measuring method

Beta ray attenuation method

Sampling filter

PTFE filter roll, L=30m (up to 60 days)

Measuring range *0 - 5 mg/ m3

Beta source *14C, less than 10 MBq.

Analog output range *0 - 0.5/ 1/ 2/ 5 mg/m3
4-ranges auto/manual switching (optional)*Starting range is selectable*Fixed range is also available

Detector

Plastic scintillation detector

Classifier *US EPA type TSP, PM10, PM2.5 or PM1

LCD display

Measuring data, Output range, performanceand control information, Messages, Alert

Internal data memory

Measuring data (1min, 1h data), Operation
history, Messages, Alert

Digital IN/OUT

Ethernet, USB1.1, RS232C

Repeatability±2% of equivalent film number

Zero drift±5μg/ m3/ day

Span drift±3%/day of equivalent film number

Linearity±5% of equivalent film number

Instrument reading forstandard air sample±10% of mass concentration

Output for serial
recorder (optional) *Measuring data (1min, 1h data), Operation
history, Messages, Alert

Instrument reading forfiltered air (blank test)±5μg/m3/day (as mean value)

Analog output for
telemeter (optional) *Stability to voltage
fluctuation for detection±3%/100±10V of equivalent film number

DC 0 - 1V : Continuous data, 1h mean value
Contact signal : Output range, external reset,
adjusting, power discontinuity, disconnecting,system failure, etc.

Power requirement

AC100V 50/ 60Hz approx. 400VA

Stability to voltage
fluctuation for flow control±5%/100±10V of setting flow rate

Withstand voltage test

AC1000V 50/ 60Hz during 1 min.

Sampling flow rate

16.7 LPM (20 C, 1atm)*

Insulation resistancemore than 5 MEG

o*For further information, please contact us.

Optional
Rack mount

A17: H875mm rollaway rack
A22: H1100mm rollaway rack

Analog I/O unit for atelemeter

TU07 : terminal block(Rear connection with rack mount)

RecorderμRP1800 serial recorder
Chart width:180mm

PM10 classifier

US EPA PM10 inlet

PM2.5 classifier

PM2.5 very sharp cut cyclone (VSCC)

PM1 classifierPM1 charp cut cyclone

Flow DiagramSample
inlet

Beta source
Sampling
head

Pressure Temperature
sensorsensorClassifier

PPTFE
filter roll

Valve for head
movementTF
Flowsensor

Detector

Flowcontroller

Sucsionpump

Dimensions
Unit: mm485

InletandTemp./RH sensor

875

485

590

230

490

Desktop type (weight : approx. 15kg)

Rack case type (weight : approx. 40kg)

Product CodePM-711640

TSP
PM10F2PM2.5

F1

PM1(3000)

TC780

Desktop type
Rack case type

S1

Outdoor case type

R

Serial recorder0No useTAnalog telemeter communication unit

0No use
85

1000

T1A1540690

Outdoor case type (weight : approx. 80kg)

CAUTION*) Operation manual should be read before measuring.
*) To avoid fire, break down and electric shock, do not use this analyzer at the place where beside water, high moisture, dust or oily smoke.

KIMOTO ELECTRIC CO., LTD.
3-1FUNAHASHI-CHO,TENNOJI-KU,OSAKA 543-0024 JAPANTEL 06-6768-3401 FAX 06-6764-7040 E-mail:sales@kimoto-electric.co.jp
http://www.kimoto-electric.co.jp
This catalog is printed out in September 2010.

*) Specifications, design and size etc. described in this catalog are subject to change without notice.

711.02