Product Information

EXE 100 Series
Interpolation andDigitizing Electronics

April 2007

EXE 100 Series
Interpolation and digitizing electronics• Input signals » 11 µAPP
• Output signals « TTLTolerancing ISO 8015
ISO 2768 - m H< 6 mm: ±0.2 mm

98

¬ 4.5

14.5

6147

 0.3 A

173

A

163±0.2

 0.3 A86±0.252±0.2

25

M4

¬ 4.5

R55

1362

Adapter plate: ID 536 452-01

7±1¬ 7.51

Accessories:
An adapter plate is available for mounting
on existing holes for the IBV 6xx/EXE 6xx:64M4*)36±0.2

Two mounting screwsM4 x 16 ISO 4762/DIN 912

21.5*)

86±0.236±0.238.5±186±0.2

 0.3 A

M4¬ 4.5

Adapter plate

M4
Md = 2.5 Nm

Included in delivery

M4

Product Information for EXE 100 Series

4/2007

Specifications

EXE 101
EXE 102

Input» 11 µAPP

Electrical connection

M23 flange socket (female) 9-pin

Cable length† 30 m for IEncoder † 120 mA1)

Interpolation

5-fold, 10-fold, 20-fold, 25-fold, 50-fold, 100-fold

Input frequency1)for interpolation

Nominal values2)

EXE 101

5-fold 100 kHz

100 kHz100 kHz100 kHz80 kHz

50 kHz25 kHz

10-fold 100 kHz

100 kHz66 kHz

50 kHz40 kHz

25 kHz12.5 kHzEXE 102 20-fold 60 kHz

50 kHz33 kHz

25 kHz

20 kHz12.5 kHz6.25 kHz

25-fold 60 kHz

40 kHz

26 kHz

20 kHz

16 kHz

10 kHz

5 kHz

50-fold 40 kHz

20 kHz

13 kHz

10 kHz

8 kHz

5 kHz

2.5 kHz

100-fold 20 kHz

10 kHz6.6 kHz

5 kHz

4 kHz

2.5 kHz1.25 kHz‡ 0.465 µs‡ 0.585 µs‡ 0.950 µs‡ 1.925 µs

Output« TTL (clocked)

Electrical connection

M23 flange socket (male) 12-pin

Cable length† 100 m (¥ † 50 m)

Edge separation a‡ 0.100 µs

Reference mark signal1)

Pulse width 90° elec. or 270° elec.

1)‡ 0.220 µs‡ 0.345 µs

Fault indication

Through fault detection signal ¥ or, in addition, Ua1/Ua2 high impedancePower supply

5 V ± 5%Current consumption3)

EXE 101: † 120 mA
EXE 102: † 140 mA

Operating temperature
Storage temperature

0 °C to 70 °C–30 °C to 80 °C

Vibration 50 to 2000 Hz
Shock 11 ms† 100 m/s2
† 300 m/s2

Degree of protection

IP 65

WeightApprox. 0.3 kg

Bold: These preferred versions are available on short notice, please select when orderingAdjustable
2)
The actual input frequency can be up to 5% lower. Exceeding this limit results in failure3)
Not including output load (80 mA with recommended input circuitry) or the current consumption of the encoder(see the corresponding brochure)1)

Product Information for EXE 100 Series

4/2007Electrical Connection

Connecting cable or adapter
cable with M23 connector(male), 9-pin

M23 connecting cable12-pin, ¬ 8 mmComplete
ID 298 399-xx«»

Cable and connector, 9-pin
See HEIDENHAIN catalogs fordigital readout and lengthgauges as well as product
Information sheets for therespective encoders

With one connector
ID 309 777-xx
Cable only
[4(2x0.14mm2) + (4x0.5 mm2)]
ID 244 957-01
Connector (female) 12-pin
ID 291 697-05

EXE input – » 11 µAPP9-pin
M23 flange socket

Power supply

Incremental signals

34Chassis

912

5678

UP0V

External
shieldInternal
shield

I1+

I1–

I2+

I2–

I0+

I0–

Brown

White–White/Brown

GreenYellow

BlueRed

GrayPink

UP = power supply voltage
Vacant pins or wires must not be used!

Shield on housing
Color assignment applies only to extension cable.

EXE output – « TTL
12-pinM23 flange socket12-pin
M23 connector

Power supply

Incremental signals

Other signals

122101156813479UPSensor
5V0VSensor
0VUa1



Ua2

£

Ua0¤

¥PWTTestpin

Brown/
GreenBlue

White/
GreenWhiteBrown

GreenGray

Pink

Red

Black

VioletYellow

Shield on housing; UP = power supply voltage
Sensor: The sensor line is connected internally with the corresponding power line

For more information• Product overview: Interface Electronics
DR. JOHANNES HEIDENHAIN GmbH
Dr.-Johannes-Heidenhain-Straße 583301 Traunreut, Germany{ +49 (86 69) 31-0
| +49 (86 69) 50 61
E-Mail: info@heidenhain.dewww.heidenhain.de624 602-21 · 10· 4/2007 · F&W · Printed in Germany · Subject to change without notice