1433D/E/F/H
Signal Generator(1MHz to 20GHz/26.5GHz/40GHz/50GHz)

Ceyear Technologies Co., Ltd.

Product Overview
1433 Series signal generator is a hand-held instrument specially designed for field testing byCeyear Technology Co., Ltd. It has the functions of CW signal output, FM/AM/Pulse modulation,
large dynamic range amplitude adjustment, step/list sweep etc. The integrated design of 8.4-inch
large capacitive touch screen is convenient for users’ operation.
1433 series signal generator has the characteristics of small size, flexible power supply andgood working environment adaptability, which is very suitable for on-site use. It can be applied to
the field installation, debugging and daily maintenance of fault diagnosis of electronic integrated
system, receiver performance test, radar, communication, navigation and other equipment test.

Product Features


Wider frequency coverage: from 1MHz to 50GHz

Higher frequency resolution: 0.1Hz



Excellent phase noise performance: -110dBc/Hz@10kHz frequency offset @10GHz
carrier(typical）

Various auxiliary test interfaces: reference input/output, pulse input, monitoring output,
synchronization output etc.

Convenient and fast user operation experience: 8.4-inch large screen with bright LCD,
convenient capacitive touch screen operation, integrated design of LCD and touch screenStrong environmental adaptability: the working temperature range is -10℃～50℃

Flexible power supply mode: can be powered by battery or power adapter

Wider frequency coverage
Thefrequency
rangeof1433
seriessignalgenerator
is1MHz
to20GHz/26.5GHz/40GHz/50GHz, the serialization minimum frequency is as low as 1MHz, and thehighest frequency is as high as 50GHz, which can meet the needs of wide-band testing.

Various auxiliary test interfaces

1

Excellent phase noise performance
1433 Series signal generator SSB phase noise is better than -110dBc/Hz@10kHz frequency
offset @10GHz carrier, which can meet the test requirements of most application scenarios.

Note 1: The supplementary features given in the form of typical values are for users' reference.

Convenient and fast user operation experience
8.4-inch large screen with high-brightness LCD, 800×600 pixel resolution, convenient capacitive
touch screen operation, multi-window display, clear display of instrument setting parameters andstatus information, providing convenient and fast user operation experience.

2Typical Application
Electronic system anti-jamming performance test
The 1433 series signal generator has a wide output frequency and power range, and has avariety of analog modulation functions. It can simulate and generate jamming signals in the actual
combat environment during the test of the anti-jamming performance of the electronic system,
which can be used for the test of the anti-jamming performance of the electronic system.

Radar reception performance test and troubleshooting
For radar and other electronic equipment receiving performance testing and troubleshooting
applications, 1433 series signal generators provide CW, analog modulated signal output, and canprovide excitation signal simulation.

Receiver under test
Demod.

Field test of antenna pattern test
For the field test application of the antenna pattern, the 1433 series signal generator outputs asignal with known fixed amplitude, which is used to test the indicator of the transmitting antennapattern.

Transmitter
Antenna

ReceiverAntenna

Signalgenerator

ReceiverControl PC3Technical Specification
1.

Main Functions2.

CW signal output function
FM/AM/Pulse modulation function
Step and List sweep function
Unstable amplitude and Unlock alarm function
Support LAN and USB interface program control function

Specifications

Frequency Characteristics
Frequency

1433D:1MHz～20GHz
Frequency Range

1433E:1MHz～26.5GHz
1433F:1MHz～40GHz
1433H:1MHz～50GHz

Frequency Resolution

N（internal YOharmonic number）

1MHz≤f＜2.35GHz

1/22.35GHz≤f＜2.5GHz

1/8

2.5GHz≤f＜5GHz

1/45GHz≤f≤10GHz

1/210GHz＜f≤20GHz

120GHz＜f≤40GHz

240GHz＜f≤50GHz

40.1Hz-6Aging rate

±0.5×10 /year

Temperature effects±0.3×10 （-10℃～50℃，versus 25℃±5℃）

Internal timebase-6-6Initial calibration accuracy ±0.5×10

frequency

1MHz～100MHz，step 1MHzPower-5dBm～+10dBm，impedance: 50Ω

Frequency

10MHzPower＞0dBm，50Ω impedance

Sweep mode

Step/List

Sweep points

2 to 1601

Dwell time

10ms to 100s

Trigger mode

Auto/manual

1MHz≤f＜2.5GHz-120dBm～+5dBm

2.5GHz≤f≤10GHz-120dBm～+10dBm

range(25℃±10℃，CW mode) 10GHz＜f≤20GHz-120dBm～+5dBm

20GHz＜f≤40GHz-120dBm～+5dBm

Reference input

Reference output
Sweep characteristics

Sweep featuresLevel characteristics

Stable output power4

40GHz＜f≤50GHz-120dBm～0dBm-10dBm＜P≤Maximum stable±1.0dB

Level accuracy

output power（25℃±10℃）-60dBm＜P≤-10dBm±1.5dB-90dBm＜P≤-60dBm±1.8dB

Output impedance

SWR

Maximum reverse power

50Ω（Rating）
1MHz≤f≤20GHz＜1.8:1

20GHz＜f≤40GHz＜2.0:140GHz＜f≤50GHz＜2.5:1+27dBm（0V DC）（Rating）

Spectral purity（specification is point frequency without modulated mode）
1MHz≤f≤1.5GHz≤-40dBc

1.5GHz＜f≤2.5GHz≤-30dBc

2.5GHz＜f≤19GHz≤-40dBc

19GHz＜f≤25GHz≤-30dBc

25GHz＜f≤50GHz≤-35dBc（typical）

1MHz≤f＜2.5GHz≤-54dBc

2.5GHz≤f＜5GHz≤-60dBc

Non-harmonics

5GHz≤f≤10GHz≤-56dBc（0dBm，>10kHz offset）

10GHz＜f≤20GHz≤-50dBc

20GHz＜f≤38GHz≤-44dBc

38GHz＜f≤50GHz≤-40dBc

Harmonics（Measured at +5dBm ormaximum specified power,
whichever is lower）≤-82dBc/Hz@100Hz≤-98dBc/Hz@1kHz

1MHz≤f＜2.35GHz≤-108dBc/Hz@10kHz≤-106dBc/Hz@100kHz≤-94dBc/Hz@100Hz≤-110dBc/Hz@1kHz

2.35GHz≤f＜2.5GHz≤-120dBc/Hz@10kHzSSB Phase noise（at≤-118dBc/Hz@100kHz

maximum stable output≤-88dBc/Hz@100Hzpower）≤-104dBc/Hz@1kHz

2.5GHz≤f＜5GHz≤-114dBc/Hz@10kHz≤-112dBc/Hz@100kHz≤-82dBc/Hz@100Hz≤-98dBc/Hz@1kHz

5GHz≤f≤10GHz≤-108dBc/Hz@10kHz≤-106dBc/Hz@100kHz
5≤-76dBc/Hz@100Hz≤-92dBc/Hz@1kHz
10GHz＜f≤20GHz≤-102dBc/Hz@10kHz≤-100dBc/Hz@100kHz≤-70dBc/Hz@100Hz≤-86dBc/Hz@1kHz

20GHz＜f≤40GHz≤-96dBc/Hz@10kHz≤-94dBc/Hz@100kHz≤-68dBc/Hz@100Hz≤-84dBc/Hz@1kHz

40GHz＜f≤50GHz≤-94dBc/Hz@10kHz≤-92dBc/Hz@100kHz

Modulation Characteristics
On/off ratio≥80dB

Rise/fall times≤30ns

Pulse modulation（Frequency >10MHz）

minimum pulse width ALC ON 1us（Deviation±50ns）
minimum pulse width ALC

100ns（Deviation±20ns）

OFFModulation typeLinear modulation、exponent modulation

Modulation rate（3dB

DC～20kHzbandwidth）
Amplitude modulation（Frequency >10MHz）

Linear closed loop：≥90%
Maximum depth
Exponent closed loop：≥20dB±（5％× setting depth＋1%）（at 1kHz modulation rate）

Linear AM accuracy±（5％× setting depth＋1dB）（at 1kHz modulation

Exponent AM accuracyrate）

Modulation rate（3dB

DC～20kHzbandwidth）
Frequency modulation（Frequency >10MHz）

N×800kHz（N: YO harmonic number），accuracy:

Maximum peak deviation±10%（at 1kHz）±3%（at 1kHz，100kHz offset，300Hz～3kHz

Distortion

demodulation bandwidth）

General characteristics
1433D

N（female），impedance: 50Ω

1433E

2.4mm（male），impedance: 50Ω

1433F

2.4mm（male），impedance: 50Ω

1433H

2.4mm（male），impedance: 50Ω

RF output port

Dimensions

314mm×218mm×91mm（W×H×D）（excluding handle, foot mat and footing）

6≤5.5kg（including battery）

Weightinput：100～240V、50/60Hz AC
Power adapter

output：15VDC，4A

Power supply
Lithium electronic battery

10.8V，9900mAh≤45W（Battery charging is not included）

Power consumption-10℃～+50℃（battery charging temperature: 0℃～

Operating temperature+45℃）

Temperature range-40℃～+70℃（battery storage temperature: -20℃～

Storage temperature+60℃）

Pulse input

BNC（male）

Synchronization output

BNC（male）

Monitoring output

BNC（male）

Reference input/output

BNC（male）

Other interface

Note: Ratings refer to expected performance, or describe product performance that is useful in the product but notcovered by the product warranty.

Ordering Information
Main Models1433D signal generator
1433E signal generator
1433F signal generator
1433H signal generator

1MHz～20GHz
1MHz～26.5GHz
1MHz～40GHz
1MHz～50GHz

Standard Configuration:
S/N

Description

Remarks
Standard three-core power cable1

Power adapter：input 100～240V

Power cable assembly

50/60Hz, output 15V/4A.
Lithium-ion rechargeable battery

2

Product quick use guide3

USB cable

4

Certificate conformity

USB remote control cable

Options：
S/N

Option ID

Description

11433-001

User Manual (Chinese)

21433-002

User Manual (English)

Function731433-003

Programming Manual (Chinese)

41433-004

Programming Manual (English)
Provide51433-S01

USB Power Meter Option(software)

Function

USBPower(RequiresMeasurement
USBPowersensor:H06~H13)
Optional Accessories of English

61433-H01

71433-H02

Power Adapter

Power Adapter

81433-H03

Rechargeable Lithium Ion Battery

Standby Battery

91433-H04

Purple Cat5e Cable

Point to Point, 2 Meters

101433-H05

Micro SD CardClass4, Capacity: 8G

111433-H06

87230 USB CW Power Sensor

121433-H0787231 USB CW Power Sensor

131433-H08

87232 USB CW Power Sensor

141433-H0987233 USB CW Power Sensor

151433-H1087234D USB Peak Power Sensor

161433-H1187234E USB Peak Power Sensor

171433-H1287234F USB Peak Power Sensor

181433-H1387234L USB Peak Power Sensor

191433-H14

Functional Bag

Protect the Instrument

201433-H15

Backpack

Easy to Carry

211433-H16

Safety Instrument Carrying Case

Version8

English Signs、Keys、Menu

9kHz ～ 6GHz ， for

CW

power

measurement（S01 is optional）
10MHz ～ 18GHz ， for

CW

powermeasurement（S01 is optional）
50MHz ～ 26.5GHz ， for CW power
measurement（S01 is optional）
50MHz ～ 40GHz ， for

CW

power

measurement（S01 is optional）
50MHz ～ 18GHz ， for Peak power
measurement（S01 is optional）
50MHz ～ 26.5GHz ， for Peak power
measurement（S01 is optional）
50MHz ～ 40GHz ， for Peak power
measurement（S01 is optional）
500MHz ～ 67GHz ， for Peak power
measurement（S01 is optional）

High strength light weight packing casewith handle for transportation