MikroscopyAnalytics

MVP DC Series
Diaphragm pumps as backing pumps
for HiPace turbopumps

Leak detection

MVP DC Series
Diaphragm pumps as backing pumps
for HiPace turbopumps

Easy system integration for one complete
vacuum solution

Your added value

Clean and dry
vacuumClean, dry vacuum
Diaphragm pumps function completely without any oil anddo not generate any particles. As a result these pumps
are ideally suited for applications in which a clean and dry vacuum is necessary. Another major benefit of diaphragmpumps is their long lifetime. At sensitive locations, diaphragmpumps not only perform reliable, but with a minimum ofvibration and noise. The noise level is usually below 50 dB(A).
Low voltage 24 V DC
The diaphragm pumps of the MVP DC series are the ideal backing pumps for turbopumps due to the 24 V DC power supply on the HiPace turbopumps. The electronic DC motor
allows adjustments to the rotational speed according tospecific applications. With the automatic boost mode, thepump will start at full speed and lower the speed once acertain level of vacuum is reached to ensure high performance, long maintenance intervals, as well as low noise andvibrations. Thanks to its stand-by mode the pump achieves aparticularly high energy efficiency.

l3

Increased processdurability
and reliabilityLess safety
precautions
due to low voltage
24 V DCLow cost ofownership

Increased mobility
due to low weight

Low maintenancefor maximum
reliability

Highly reliable
The 2-stage pumps are equipped with a magnetic gas ballast
valve for increased process reliability and durability. Operating
temperature ranges from 5 to 40 °C to cover a wide range ofapplications. With their easy maintenance and long service intervals of approx. 2 years, these pumps guarantee very low
cost of ownership.
Corrosive gas version
Our MVP 030-3 C DC is characterised by a gas ballast valve
and very good chemical and condensate compatibility.
The special choice of materials ensures significantly increased
robustness in a wide range of processes.
Applications■ Mass spectrometry■ Electron microscopy■ Residual gas analysis■ Surface analysis■ Leak detection■ Pumping stations■ R&D experiments■ X-Ray diffraction

MVP DC Series
Pumping speed, dimensions
Pumping speed
MVP 010-3 DC

MVP 015-2 DC

MVP 010-3 DC

MVP 015-2 DC

S [h3/h]

10S [m3/h]

1.03000 min-1

1800 min-1

2600 min-1

11800 min-1
0.1

0.1

0.010.1

110100101001000

p [hPa]p [hPa]MVP 030-3 DC

1.0

1800 minS [m3/h]MVP 015-4 DC
MVP 015-4 DC
S [m3/h]

0.01
11000-1

1500 min-1

10MVP 030-3 DC

2400 min-1
1800 min-1

0.1

0.0

1.00.11101001000

p [hPa]0.1

1101001000

p [hPa]MVP 030-3 C DC

MVP 030-3 C DC

S [m³/h]

102400 min

1800 min10.11

10100

1000

p [hPa]Nominal rotation speed
Boost modetion
fficient due
Easy installa
sy and cost-eearyve. Justisup
15-pin D-sub
System setunication viamCUmD
cop,
ed
umifi
e turbopto the unwith a HiPac
aPVve
M
haeu
thyot
connecsupply anderw
ze
poS
ct
TPdd compa sicontroller anw weight an
lor
eir
Th.
folem
ea
id
running syst
VP DC series
mps of the Mal systems
icyt
alanl
make the pual
smlyngsieaincrobile
well as for m
integration inng stations aspium
oprbtuandapplications.
on page 7)(See graphicl5Dimensions
MVP 010-3 DCMVP 015-2 DC208
180

140

105
45

172.5

45

62

109

99
V

31

G1/8"

46

47.5

22

22
6.3

G1/8"

G1/8"

142

99

75
37.5.4
ø435

G

ø8/6
V

G1/8"

72

62

57

21.5

12062
100

MVP 015-4 DC

MVP 030-3 DC
59
G1/8"

VG1/8"

62

147.7

149
163

217

22160
230

V

1045

141

G1/8"

96
140

264

117

silencer
G1/8"

126

Easy installation

35223

143

MVP 030-3 C DCTPSDCUMVP

175

156

223

PTFE tube 10/8HiPaceCable
PE 100 013 -T

116
142

PTFE tube
10/8121238

Dimensions in mm

MVP DC Series
Technical data, order numbers,accessories, spare parts
Technical data
Flange (out)Flange (in)Exhaust pressure, max
Rotation speed
Rotation speed max.
Ultimate pressure with gas ballast
Ultimate pressure without gasballast
Weight with motor
Cooling method, standard
Pumping speed, max.
Protection category
Current consumption, max
Ambient temperature
Supply: Voltage
Emission sound pressure level
without gas ballast
Certification

MVP 010-3 DC

MVP 015-2 DCMVP 015-4 DC

G 1/8“with silencer
G 1/8“
1,100 hPa
1,800 1/min
3,000 1/min–G 1/8“with silencer
G 1/8“
1,100 hPa
1,800 1/min
2,600 1/min
4 hPa

G 1/8“with silencer
G 1/8“
1,100 hPa
1,500 1/min
1,800 1/min–

1.0 hPa

3.5 hPa

0.5 hPa

2.0 kgAir
0.6 m3/h
IP 20
2.5 A
5 – 40 °C
24 (± 10%) V DC

2.7 kgAir
1.0 m3/h
IP 20
4.6 A
5 – 40 °C
24 (± 10%) V DC

4.4 kgAir
0.75 m3/h
IP 20
6.8 A
5 – 40 °C
24 (± 10%) V DC

50 dB (A)50 dB (A)45 dB (A)cTUVus certified
according to UL /CSA 61010-1

cTUVus certified
according to UL /CSA 61010-1

cTUVus certified
according to UL /CSA 61010-1

MVP 010-3 DC
PK T05 072

MVP 015-2 DC
PK T05 150

MVP 015-4 DC
PK T05 160

Order numbers
Diaphragm pump

Technical data
MVP 030-3 C DC

1.8 m3/h
Pumping speed

45 dB (A)Emission sound pressure level
without gas ballast

Flange (out)Flange (in)Exhaust pressure, max
Rotation speed
Rotation speed max.
Ultimate pressure with gas ballastUltimate pressure without gas
ballastWeight with motor
Cooling method, standard
Pumping speed, max.
Protection category
Current consumption, max
Ambient temperature
Supply: Voltage
Emission sound pressure level
without gas ballast
Certification

MVP 030-3 DC

MVP 030-3 C DC

G 1/8“with silencer
G 1/8“
1,100 hPa
1,800 1/min
2,400 1/min
3.0 hPa

G 1/8“with silencer
G 1/8“
1,100 hPa
1,800 1/min
2,400 1/min
4.5 hPa

2.0 hPa

2.5 hPa

4.3 kgAir
1.8 m3/h
IP 20
7A
10 – 40 °C
24 (± 10%) V DC

4.3 kgAir
1.8 m3/hIP 20
7A
10 – 40 °C
24 (± 10%) V DC

45 dB (A)45 dB (A)cTUVus certified
according to UL /CSA 61010-1

cTUVus certified
according to UL /CSA 61010-1

MVP 030-3 DC
PK T01 500

MVP 030-3 C DC
PK T01 600

Order numbers
Diaphragm pumpl7
Accessories
MVP 010-3 DC
MVP 015-2 DC
Power supply
TPS 110, mains pack for wall/standard rail fitting
TPS 180, mains pack for wall/standard rail fitting
TPS 111, mains pack 19“ rack module 3HU
TPS 181, mains pack 19“ rack module 3HUPower supplies and control displays
DCU 110, Display Control Unit incl. power supply
DCU 180, Display Control Unit incl. power supply
DCU 002, Display Control Unit
HPU 001, Handheld Programming Unit
CablesConnecting cable MVP-TC-TPS (size 3 m)Connecting cable supply 24V/48Vto drive electronics with RS-485 interface
USB converter to RS-485 interface
Y-Connector M12 to RS-485
Interface cable, M12 m straight/M12 m straight, 3 m
Flange and hose connection
Screw-in flange DN 16 ISO-KF / G 1/8“ incl. seal
Push-in fitting G 1/8 – 8/6
Hose DN 6; (ø 8/6) Polyurethane
Hose connection DN 6 x 400 mm
for MVP and HiPace turbopumps
Hose connection DN 6 x 1000 mm
for MVP and HiPace turbopumps
Protection
Purge gas nozzle with filter for MVP 030-3 DC

MVP 030-3 DC
MVP 030-3 C DC
MVP 015-4 DC

PM 061 340 -T
PM 061 341 -T
PM 061 344 -T
PM 061 345 -T

PM C01 820
PM C01 821
PM 061 348 -T
PM 051 510 -T

PE 100 013 -T
PM 061 350 -T
PM 061 207 -T
P 4723 010
PM 061 283 -T

PK 050 108 -T
P 4131 029 GP 0991 939
P 0920 739 E
P 0920 817 EPK 050 304 -U

Spare parts
Order numbers
Set of spare parts MVP 010-3 DC
Set of spare parts MVP 015-2 DC
Set of spare parts MVP 015-4 DC
Set of spare parts MVP 030-3 DC
Set of spare parts MVP 030-3 C DC
Silencer G 1/8“PU E22 011 -T
PU E22 001 -T
PU E22 031 -T
PU E22 003 -T
PU E22 032 -T
P 0920 567 E

All information is subject to change without prior notice. PU 0064 PEN (May 2021/PoD)Are you looking for youroptimized vacuum solution?
Please contact us:

Pfeiffer Vacuum GmbHGermany
T +49 6441 802-0

www.pfeiffer-vacuum.com