MVP 010-3 DC, diaphragm pump,24 V DC

MVP 010-3 DC, diaphragm pump, 24 V DC
Compact diaphragm pump with a pumping speed of up to 0.6 m3/h
Pumping of dry and non-corrosive gases
Industry 4.0 compatible: Operation with other intelligent products, such as turbopumps
via RS-485
Similar Image

Dimensions

www.pfeiffer-vacuum.com

Page 2

Technical Data

MVP 010-3 DC, diaphragm pump, 24 V DC

Pumping speed
Final pressure without gas ballast
Leakage rate

0.6 m³/h | 0.35 cfm | 10 l/min
1 hPa | 0.75 Torr | 1 mbar≤ 5 · 10-4 Pa m3/s | ≤ 3.75 · 10-3 Torr l/s |≤ 5 · 10-3 mbar l/s
1,100 hPa | 825 Torr | 1,100 mbar
1,300 hPa | 975 Torr | 1,300 mbar
G 1/8"
1/8" thread with silencer (separate package)
RS-48524 V DC
2.5 A
540-3,000 rpm | 540-3,000 min-1
1,800 rpm | 1,800 min-1
cTUVus certified according to UL / CSA 610101
IP20No50 dB(A)
5-40 °C | 41-104 °F | 278-313 K
5-40 °C | 41-104 °F | 278-313 K-10-60 °C | 14-140 °F | 263-333 K
Convection cooled
2 kg | 4.41 lb
No2000 mIntake pressure max.
Exhaust pressure, max.
Connection flange (in)Connection flange (out)I/O interfaces
Supply: Voltage V DC
Current, max.
Rotation speed
Nominal rotation speed
Certification
Protection degree
SwitchSound pressure level
Ambient temperature
Temperature: Storage
Temperature: Shipping
Cooling method
WeightGas ballast
Operating altitude, max.
Order number
MVP 010-3

www.pfeiffer-vacuum.com

PK T05 072

Page 3

Accessories
Adapters (electrical)|RS-232/485
USB RS-485 converter
Y-connector M12 for RS-485
Cables|Backing pump control cablesConnection cable MVP-TC-TPS, 3 m
Cables|Connection cables
Connection cable for HiPace with TC 110/120
Connecting cable from 24V/48V power supply
pack to electronic drive unit. With RS-485
interface
Cables|RS-485 Interface cables
Interface cable, M12 m straight /M12 m straight, 3 m
Connections|Adapters Turbopump toBacking Pump
Hose DN 6; (ø 8/6 mm) polyethylene
Hose connection DN 6 x 400 mm with straightconnector 1/8" thread and 1/4" thread including
seal
Hose connection DN 6 x 1000 mm with straightconnector 1/8" thread and 1/4" thread includingseal
Push-in T-fitting 1/8" thread including seal forhose connection (8/6 mm)
Push-in fitting 1/8" thread including seal forhose connection (8/6 mm)
Connections|Screw-in flanges
Screwing flange DN 16 ISO-KF / 1/8" threadincl. seal
Control Devices|Power supplies and control
displays
OmniControl 001, rack unit without integrated
power supply pack
OmniControl 200, rack unit with integrated
power supply packTPS 110, power supply pack for wall/standard
rail installation
TPS 111, power supply pack 19" plug-in unit
3HU
Flanges/Ports|Blank flanges
Push-in fitting G 1/8" including seal for hoseconnection (8/6 mm)Monitoring|Pressure Sensors
OmniControl 200 with TPR 270
Vibration Reduction|Foot plates
Foot plate with rubber feetwww.pfeiffer-vacuum.com

PM 061 207 -T
P 4723 010
PE 100 013 -T
PM 061 543 -T

PM 061 350 -T

PM 061 283 -T

P 0991 939

P 0920 739 E

P 0920 817 E
P 4131 030 E
P 4131 026 C

PK 050 108 -T

PE D40 000 0
PE D50 000 0
PM 061 340 -T
PM 061 344 -T

P 4131 029 G
PT 440 950 -T
PK 050 273

Page 4

Vacuum solutions from a single source

Pfeiffer Vacuum stands for innovative and custom vacuum solutions worldwide,
technological perfection, competent advice and reliable service.

Complete range of products

From a single component to complex systems:
We are the only supplier of vacuum technology that provides a complete product portfolio.

Competence in theory and practice

Are you looking for aperfect vacuum solution?
Please contact us:

www.pfeiffer-vacuum.com

Pfeiffer Vacuum GmbH
Headquarters · Germany
T +49 6441 802-0
info@pfeiffer-vacuum.deAll data subject to change without prior notice.

Benefit from our know-how and our portfolio of training opportunities!
We support you with your plant layout and provide first-class on-site service worldwide.