MVP 015-2 DC, diaphragm pump,
24 V DCMVP 015-2 DC, diaphragm pump, 24 V DC

Similar Image

Diaphragm pump with a pumping speed of up to 1.0 m3/h
No hydrocarbons: Completely dry &amp; oil-free diaphragm pump
Low noise level, low vibrations and compact design, also ideal for use in laboratories
Low operating costs thanks to the highest quality, short maintenance times and long
maintenance intervals
Industry 4.0 compatible: Operation with other intelligent products, such as turbopumps
via RS-485

Dimensions

www.pfeiffer-vacuum.com

Page 2Technical Data

MVP 015-2 DC, diaphragm pump, 24 V DC

Pumping speed
Final pressure without gas ballast
Final pressure with gas ballast
Leakage rate

1 m³/h | 0.59 cfm | 16.67 l/min
3.5 hPa | 2.62 Torr | 3.5 mbar
6 hPa | 4.5 Torr | 6 mbar
≤ 5 · 10-4 Pa m3/s | ≤ 3.75 · 10-3 Torr l/s |≤ 5 · 10-3 mbar l/s
1,100 hPa | 825 Torr | 1,100 mbar
1,100 hPa | 825 Torr | 1,100 mbar
G 1/8"1/8" thread with silencer
RS-48524 V DC
4.6 A
540-2,600 rpm | 540-2,600 min-1
1,800 rpm | 1,800 min-1
cTUVus certified according to UL / CSA 610101
IP20No50 dB(A)5-40 °C | 41-104 °F | 278-313 K
5-40 °C | 41-104 °F | 278-313 K-10-60 °C | 14-140 °F | 263-333 K
Convection cooled
2.7 kg | 5.95 lb
Yes
2000 mIntake pressure max.
Exhaust pressure, max.
Connection flange (in)Connection flange (out)I/O interfaces
Supply: Voltage V DC
Current, max.
Rotation speed
Nominal rotation speed
Certification
Protection degree
SwitchSound pressure level
Ambient temperature
Temperature: Storage
Temperature: Shipping
Cooling method
WeightGas ballast
Operating altitude, max.
Order number
MVP 015-2 DC

www.pfeiffer-vacuum.com

PK T05 150

Page 3

Accessories
Adapters (electrical)|RS-232/485
USB RS-485 converter
Y-connector M12 for RS-485
Cables|Backing pump control cables
Connection cable MVP-TC-TPS, 3 m
Cables|Connection cables
Connection cable for HiPace with TC 110/120
Connecting cable from 24V/48V power supply
pack to electronic drive unit. With RS-485
interface
Cables|RS-485 Interface cables
Interface cable, M12 m straight /M12 m straight, 3 m
Connections|Adapters Turbopump toBacking Pump
Push-in T-fitting 1/8" thread including seal forhose connection (8/6 mm)
Hose DN 6; (ø 8/6 mm) polyethylene
Hose connection DN 6 x 400 mm with straightconnector 1/8" thread and 1/4" thread including
seal
Push-in fitting 1/8" thread including seal forhose connection (8/6 mm)
Hose connection DN 6 x 1000 mm with straightconnector 1/8" thread and 1/4" thread including
seal
Connections|Screw-in flanges
Screwing flange DN 16 ISO-KF / 1/8" threadincl. seal
Control Devices|Power supplies and control
displaysTPS 110, power supply pack for wall/standard
rail installation
TPS 111, power supply pack 19" plug-in unit
3HU
OmniControl 001, rack unit without integrated
power supply pack
OmniControl 200, rack unit with integrated
power supply pack
Flanges/Ports|Blank flanges
Push-in fitting G 1/8" including seal for hoseconnection (8/6 mm)Monitoring|Pressure Sensors
OmniControl 200 with TPR 270

www.pfeiffer-vacuum.com

PM 061 207 -T
P 4723 010
PE 100 013 -T
PM 061 543 -T

PM 061 350 -T

PM 061 283 -T

P 4131 030 EP 0991 939P 0920 739 E
P 4131 026 C

P 0920 817 E

PK 050 108 -T

PM 061 340 -T
PM 061 344 -T
PE D40 000 0
PE D50 000 0

P 4131 029 G
PT 440 950 -T

Page 4

Vacuum solutions from a single source

Pfeiffer Vacuum stands for innovative and custom vacuum solutions worldwide,
technological perfection, competent advice and reliable service.

Complete range of products

From a single component to complex systems:
We are the only supplier of vacuum technology that provides a complete product portfolio.

Competence in theory and practice

Are you looking for aperfect vacuum solution?
Please contact us:

www.pfeiffer-vacuum.com

Pfeiffer Vacuum GmbHHeadquarters · Germany
T +49 6441 802-0
info@pfeiffer-vacuum.deAll data subject to change without prior notice.

Benefit from our know-how and our portfolio of training opportunities!
We support you with your plant layout and provide first-class on-site service worldwide.