GP01158D/06/EN/02.22-00
71575137
2022-08-01

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Proline Promag 10
Electromagnetic flowmeter
Modbus RS485

Services

Proline Promag 10 Modbus RS485

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

5.2

1.1
1.2
1.3

5.3

1.4

Document function . . . . . . . . . . . . . . . . . . . . .
Target group . . . . . . . . . . . . . . . . . . . . . . . . . .
Using this document . . . . . . . . . . . . . . . . . . . .
1.3.1 Symbols . . . . . . . . . . . . . . . . . . . . . . .
1.3.2 Information on the document
structure . . . . . . . . . . . . . . . . . . . . . . .
1.3.3 Structure of a parameter description . .
Associated documentation . . . . . . . . . . . . . . . .

2"Guidance" menu . . . . . . . . . . . . . . . . . . . 6

2.1"Commissioning" wizard . . . . . . . . . . . . . . . . . . 6
2.1.1 Device identification . . . . . . . . . . . . . . 8
2.1.2 System units . . . . . . . . . . . . . . . . . . . . 9
2.1.3 Totalizer 1 to n . . . . . . . . . . . . . . . . . 13
2.1.4 Process parameters . . . . . . . . . . . . . . 15
2.1.5 Measurement conditions . . . . . . . . . . 16
2.1.6 Current output . . . . . . . . . . . . . . . . . 20
2.1.7 Display . . . . . . . . . . . . . . . . . . . . . . . 25
2.1.8 Date/time . . . . . . . . . . . . . . . . . . . . . 27

4
4
4
4
4
5
5

3

"Diagnostics" menu . . . . . . . . . . . . . . . . 29

3.1
3.2
3.3
3.4
3.5

"Actual diagnostics" submenu . . . . . . . . . . . . . 30
"Diagnostic list" submenu . . . . . . . . . . . . . . . . 32
"Simulation" submenu . . . . . . . . . . . . . . . . . . . 35
"Heartbeat Technology" submenu . . . . . . . . . . 38
"Diagnostic settings" submenu . . . . . . . . . . . . . 39
3.5.1 "Properties" submenu . . . . . . . . . . . . . 39
3.5.2 "Diagnostic configuration" submenu . . 39

4"Application" menu . . . . . . . . . . . . . . . . 49

4.1

"Measured values" submenu . . . . . . . . . . . . . . 49
4.1.1 "Totalizer" submenu . . . . . . . . . . . . . . 51
"System units" submenu . . . . . . . . . . . . . . . . . 53
"Totalizers" submenu . . . . . . . . . . . . . . . . . . . 58
4.3.1 "Totalizer handling" submenu . . . . . . . 58
4.3.2 "Totalizer 1 to n" submenu . . . . . . . . . 58
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . . 63
4.4.1 "Process parameters" submenu . . . . . . 63
4.4.2 "Low flow cut off" submenu . . . . . . . . 66
4.4.3 "Partially filled pipe detection"
submenu . . . . . . . . . . . . . . . . . . . . . . 67
4.4.4 "Sensor adjustment" submenu . . . . . . . 69
4.4.5 "Calibration" submenu . . . . . . . . . . . . 70
4.4.6 "Electrode cleaning cycle" submenu . . . 72
"Current output" submenu . . . . . . . . . . . . . . . . 74
" Communication" submenu . . . . . . . . . . . . . . 85
4.6.1 "Modbus configuration" submenu . . . . 85
4.6.2 "Modbus data map" submenu . . . . . . . 88
4.6.3 "Modbus information" submenu . . . . . 89

4.2
4.3
4.4

4.5
4.6

5

"System" menu . . . . . . . . . . . . . . . . . . . . 90

5.1

"Device management" submenu . . . . . . . . . . . . 91

Endress+Hauser

5.4
5.5

5.6
5.7

"User management" submenu . . . . . . . . . . . . . 93
5.2.1 "Define access code" wizard . . . . . . . . . 95
"Connectivity" submenu . . . . . . . . . . . . . . . . . 96
5.3.1 "Bluetooth configuration" submenu . . . 96
" Date / Time" submenu . . . . . . . . . . . . . . . . . 97
"Information" submenu . . . . . . . . . . . . . . . . . . 99
5.5.1 "Device" submenu . . . . . . . . . . . . . . . . 99
5.5.2 "Sensor electronic module (ISEM)"
submenu . . . . . . . . . . . . . . . . . . . . . 102
5.5.3 "Display module" submenu . . . . . . . . 102"Display" submenu . . . . . . . . . . . . . . . . . . . . 104"Software configuration" submenu . . . . . . . . . 108

6

Modbus RS485 register
information . . . . . . . . . . . . . . . . . . . . . . 109

6.1

Notes . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 109
6.1.1 Structure of the register information 109
6.1.2 Address model . . . . . . . . . . . . . . . . 109
Overview of the operating menu . . . . . . . . . . 110
Register information . . . . . . . . . . . . . . . . . . 119
6.3.1 "Guidance" menu . . . . . . . . . . . . . . . 119
6.3.2 "Diagnostics" menu . . . . . . . . . . . . . . 125
6.3.3 "Application" menu . . . . . . . . . . . . . 128
6.3.4 "System" menu . . . . . . . . . . . . . . . . . 136

6.2
6.3

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 141

3

About this document

Proline Promag 10 Modbus RS485

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the operating
menus.
It is used to perform tasks that require detailed knowledge of the function of the device:• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Symbols

Types of information
Preferred procedures, processes or actions
Permitted procedures, processes or actions
Forbidden procedures, processes or actions
Additional information
Reference to documentationA

Reference to page
Reference to graphic

1.3.2

Information on the document structure

The parameters of all the operating menus and the commissioning wizard are described inthis document.
• Guidance menu with the Commissioning wizard (→  6), which guides the user
automatically through all the device parameters that are required for commissioning• Application menu (→  49)• Diagnostics menu (→  29)• System menu (→  90)

4

Endress+Hauser

Proline Promag 10 Modbus RS485

1.3.3

About this document

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter nameWrite-protected parameter =Navigation

Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Selection

List of the individual options for the parameter• Option 1• Option 2

User entry

Input range for the parameter

User interface

Display value/data for the parameter

Additional information

Additional explanations (e.g. in examples):• On individual options• On display values/data• On the input range• On the parameter function

1.4

Associated documentation

Technical Information

Overview of the device with the most important technical data.

Operating Instructions

All the information that is required in the various phases of the life cycle ofthe device: from product identification, incoming acceptance and storage, tomounting, connection, operation and commissioning through totroubleshooting, maintenance and disposal as well as the technical data anddimensions.

Sensor Brief Operating Instructions

Incoming acceptance, transport, storage and mounting of the device.

Transmitter Brief Operating
Instructions

Electrical connection and commissioning of the device.

Description of Parameters

Detailed explanation of the menus and parameters.

Safety Instructions

Documents for the use of the device in hazardous areas.

Special Documentation

Documents with more detailed information on specific topics.

Installation Instructions

Installation of spare parts and accessories.

The related documentation is available online:
W@M Device Viewer

On the www.endress.com/deviceviewer website, enter the serial number ofthe device: nameplate

Endress+Hauser Operations App‣ Scan the Data Matrix code: nameplate‣ Enter the serial number of the device: nameplate

Endress+Hauser

5"Guidance" menu

Proline Promag 10 Modbus RS485

2"Guidance" menu

Main functions for use – from fast and safe commissioning to guided support during
operation.
Navigation

 Guidance

Guidance‣ Commissioning

2.1

→ 6

"Commissioning" wizard

Complete this wizard to commission the device. NOTE: If you exit the wizard beforehand,the changes you made will be saved. For this reason, the device may be in an undefinedstate! In this case, reset the device to the default settings.
Navigation

 Guidance → Commissioning‣ Commissioning

6

Device tag→ 8Serial number→ 8

Firmware version→ 8

Device name→ 8Volume flow unit

→ 9Volume unit

→  10Mass flow unit

→  11Mass unit→  11

Density unit

→  12Temperature unit

→  12

Conductivity unit

→  12

Assign process variable→  13

Unit totalizer 1 to n

→  13

Totalizer operation mode→  14Endress+Hauser

Proline Promag 10 Modbus RS485

Endress+Hauser"Guidance" menu

Failure mode→  15

Flow damping→  15

Low flow cut off→  17

On value low flow cutoff→  18Off value low flow cutoff→  18Pressure shock suppression→  16

Empty pipe detection→  19Empty pipe adjust value→  19Full pipe adjust value→  19Process variable current output→  20Current range output→  20Lower range value output→  21

Upper range value output→  22

Damping current output→  23Fixed current→  23

Failure behavior current output→  23

Failure current→  24

Value 1 display→  25

Value 2 display→  25

Value 3 display→  26

Value 4 display→  26

Display damping→  26Time format→  27

Time zone→  27

Set date/time→  28

7

"Guidance" menu

Proline Promag 10 Modbus RS485

2.1.1
Navigation

Device identification
 Guidance → Commissioning

Device tagNavigation

 Guidance → Commissioning → Device tag

Description

Enter a unique designation for the measuring point to be able to easily identify it within
the plant.

User entry

Character string comprising numbers, letters and special characters (32)Serial number
Navigation

 Guidance → Commissioning → Serial number

Description

Displays the serial number of the measuring device. The serial number can be used to
identify the measuring device and to retrieve further information on the measuring device,
such as the related documentation, via the Device Viewer or Operations app.
Additional information:
The serial number can also be found on the nameplate of the sensor and transmitter.

User interface

Character string comprising numbers, letters and special characters

Firmware version
Navigation

 Guidance → Commissioning → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

Device name
Navigation

 Guidance → Commissioning → Device name

Description

Displays the name of the transmitter.
Additional information:
The name can also be found on the transmitter's nameplate.

8

Endress+Hauser

Proline Promag 10 Modbus RS485

User interface"Guidance" menu

Character string comprising numbers, letters and special characters

2.1.2
Navigation

System units Guidance → Commissioning

Volume flow unitNavigation

 Guidance → Commissioning → Volume flow unit

Description

Select volume flow unit.

Endress+Hauser

9"Guidance" menu

Selection

Proline Promag 10 Modbus RS485

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dUS units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• MMft³/s• MMft³/min• MMft³/h• Mft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)Volume unitNavigation

 Guidance → Commissioning → Volume unit

Description

Select volume unit.

10

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)Endress+Hauser

Proline Promag 10 Modbus RS485

Selection

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega"Guidance" menu

US units• af• ft³• Mft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;beer)• bbl (us;tank)

Mass flow unitNavigation

 Guidance → Commissioning → Mass flow unit

Description

Select mass flow unit.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

Mass unitNavigation

 Guidance → Commissioning → Mass unit

Description

Select mass unit.

Selection

SI units
•g
• kg
•t

Endress+Hauser

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

US units• oz• lb• STon11

"Guidance" menu

Proline Promag 10 Modbus RS485

Density unitNavigation

 Guidance → Commissioning → Density unit

Description

Select density unit.

Selection

SI units• g/cm³• g/m³• kg/l• kg/dm³• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

US units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)Temperature unitNavigation

 Guidance → Commissioning → Temperature unit

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Select temperature unit.

Selection

SI units• °C•K

US units• °F• °R

Conductivity unitNavigation

 Guidance → Commissioning → Conductiv. unit

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

12

Select conductivity unit.

Endress+Hauser

Proline Promag 10 Modbus RS485

Selection"Guidance" menu

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/m
• kS/m• MS/m2.1.3

Totalizer 1 to n

Navigation

 Guidance → Commissioning

Assign process variable

Navigation

 Guidance → Commissioning → Assign variable

Description

Select process variable for totalizer.
Additional information:
If the option selected is changed, the device resets the totalizer to "0".

Selection• Off• Volume flow
• Mass flow

Unit totalizer 1 to n

Navigation

 Guidance → Commissioning → Unit totalizer 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select process variable totalizer unit.

Selection

SI units• g*• kg *• t**

US units• oz *• lb *• STon *Visibility depends on order options or device settingsorEndress+Hauser

13

"Guidance" menu

Proline Promag 10 Modbus RS485

SI units• cm³ *
• dm³ *
• m³ *• ml *• l*• hl *• Ml Mega **US units• af *• ft³ *• Mft³ *• fl oz (us) *• gal (us) *• kgal (us) *• Mgal (us) *• bbl (us;liq.) *• bbl (us;beer) *• bbl (us;oil) *• bbl (us;tank) *Imperial units• gal (imp) *• Mgal (imp) *• bbl (imp;beer) *• bbl (imp;oil) *Visibility depends on order options or device settingsorOther units
None **

Additional information

Visibility depends on order options or device settings

Description
The unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  53).
Selection
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  13).

Totalizer operation modeNavigation

 Guidance → Commissioning → Operation mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select totalizer calculation mode.

Selection• Net flow total• Forward flow total• Reverse flow total

Additional information

Selection• Net flow total option
The flow values in the forward and reverse flow directions are totalized and netted
against each other. Net flow is recorded in the flow direction.
• Forward flow total option
Only the flow in the forward flow direction is totalized.
• Reverse flow total option
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

14

Endress+Hauser

Proline Promag 10 Modbus RS485"Guidance" menu

Failure modeNavigation

 Guidance → Commissioning → Failure mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify how the totalizer should behave in the event of a device alarm.
Additional information:
The failsafe mode that applies to any other totalizers or outputs is specified separately inother parameters and is not impacted by this setting.

Selection• Stop• Actual value• Last valid value

Additional information

Selection• Stop option
The totalizer is stopped in the event of a device alarm.
• Actual value option
The totalizer continues to totalize based on the current value measured; the device alarmis ignored.
• Last valid value option
The totalizer continues to totalize based on the last valid value measured before thedevice alarm occurred.

2.1.4
Navigation

Process parameters Guidance → Commissioning

Flow damping

Navigation

 Guidance → Commissioning → Flow damping

Description

Enter value for damping of the flow measured value in order to reduce the variability of
the flow measured value when exposed to interference.
Additional information:
The depth of the flow filter is determined by this setting. As the filter depth increases, sodoes the reaction time of the device.
- Value = 0: No damping. Damping of 0 is not recommended, as the measuring signal isthen so noisy that it is almost impossible to perform a measurement.
- Value > 0: Damping increases
Optimal damping depends on the measuring period.
Damping impacts the following measuring device variables:- Outputs- Low flow cut off- Totalizers

Endress+Hauser

15

"Guidance" menu

User entry

Proline Promag 10 Modbus RS485

0 to 15

2.1.5

Measurement conditions

Pressure shock suppression
Navigation

 Guidance → Commissioning

Pressure shock suppression

Navigation

 Guidance → Commissioning → Pres. shock sup.

Description

Enter time frame for signal suppression (= pressure shock suppression active), for example
to prevent the device from registering flow movements in the pipe when a valve is closed.
Additional information:
Pressure schock suppression is activated when the flow rate drops below the on value forlow flow cutoff. Output values when pressure shock suppression is active:- Flow: 0- Totalizer: Last valid value
Pressure shock suppression is deactivated when:- The time frame specified has elapsed and- Flow exceeds the off value for low flow cutoff

User entry

0 to 100 s

Additional information

ExampleWhen a valve is closed, momentarily strong fluid movements may occur in the pipeline,
which are registered by the device. These totalized flow values lead to a false totalizer
status, particularly during batching processes.

16

Endress+Hauser

Proline Promag 10 Modbus RS485"Guidance" menu

C

D

Q

2

1

D

3

4

6
5

t

A

B
A0012888

Q
t
A
B
C
D
1
2
3
4
5
6

Flow
Time
After runPressure shock
Pressure shock suppression active as specified by the time entered
Pressure shock suppression inactive
Valve closes
Flow falls below the on-value of the low flow cut off: pressure shock suppression is activated
The time entered has elapsed: pressure shock suppression is deactivated
The actual flow value is now displayed and output
On-value for low flow cut off
Off-value for low flow cut off

Low flow cut off
Navigation

 Guidance → Commissioning

Low flow cut offNavigation

 Guidance → Commissioning → Low flow cut off

Description

Select process variable for low flow cut off to activate low flow cut off.

Selection• Off• Volume flow
• Mass flow

Additional information

Description

Endress+Hauser

17

"Guidance" menu

Proline Promag 10 Modbus RS485

Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn-value entered
Off-value entered

On value low flow cutoff

Navigation

 Guidance → Commissioning → On value

Description

Enter on value to switch on low flow cut off.
Value = 0: No low flow cut off
Value > 0: Low flow cut off is activatedUser entry

Positive floating-point number

Off value low flow cutoff

Navigation

 Guidance → Commissioning → Off value

Description

Enter off value to switch off low flow cut off. The off value is entered as a positive
hysteresis with respect to the on value.

User entry

0 to 100.0 %18

Endress+Hauser

Proline Promag 10 Modbus RS485"Guidance" menu

Empty pipe detection
Navigation

 Guidance → Commissioning

Empty pipe detection



Navigation

 Guidance → Commissioning → Empty pipe det.

Description

Switch empty pipe detection on or off. Switch on empty pipe detection to detect a partiallyfilled or empty measuring tube.

Selection

• Off
• On

Empty pipe adjust value

Navigation

 Guidance → Commissioning → Empty pipe value

Description

Displays adjustment value when the measuring tube is empty.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

Full pipe adjust value

Navigation

 Guidance → Commissioning → Full pipe value

Description

Displays adjustment value when the measuring tube is full.
NOTE
Users logged on in the Service role have write access!

User interface

Endress+Hauser

Positive floating-point number

19

"Guidance" menu

Proline Promag 10 Modbus RS485

2.1.6

Current output

Navigation

 Guidance → Commissioning

Process variable current output
Navigation

 Guidance → Commissioning → Proc.var. outp

Description

Select process variable for current output

Selection• Off *• Mass flow
• Conductivity *• Corrected conductivity *• Volume flow
• Temperature *• Noise *• Coil current shot time *Current range output

Navigation

 Guidance → Commissioning → Curr.range out

Description

Select the current range for the measured value output and the upper and lower fault
condition signal level.
Additional information:- The measured value range is specified in the "Lower range value output " parameter andthe "Upper range value output " parameter.
- If the measured value lies outside the scaled measured value range, diagnostic message"441 Current output faulty" is generated.
- In the event of a device alarm, the current output adopts the behavior specified in the"Failure behavior current output " parameter.

Selection• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)• Fixed value

Additional information

Selection• 4...20 mA NE (3.8...20.5 mA) option
Select this option to set the current range in accordance with NAMUR recommendation
NE43.
• Fixed value option
Select this option to set the current output to a current value instead of a range.
The current value is defined in the Fixed current parameter (→  23).

*

20

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485"Guidance" menu

The graphic shows the relationship between the current range for the output of theprocess value and the lower and upper alarm levels:
2

1

3

I [mA]
A0034351

1
2
3

Current range for process value
Lower level for signal on alarm
Upper level for signal on alarm

Selection (current range for process value) Lower level for signal on
alarm

Upper level for signal on
alarm

4...20 mA NE (3.8...20.5 mA)> 21.5 mA< 3.6 mA

4...20 mA US (3.9...20.8 mA)
4...20 mA (4... 20.5 mA)

Lower range value output

Navigation

 Guidance → Commissioning → Low.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter lower range value for the measured value range.
Additional information:- Depending on the setting selected for the "Measuring mode current output " parameter,
the value specified for this parameter and the "Upper range value output " parameter musthave the same algebraic sign or not.
- As a rule, the lower range value is scaled to be lower than the upper range value. As aresult, the behavior of the current output is proportional to the process variable assigned.
If the lower range value is scaled to be higher than the upper range value, then the
behavior of the current output will be inversely proportional to the process variableassigend.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter (→  75).
Example: Measuring mode with "Forward flow" option• Lower range value output parameter (→  21) = not equal to zero flow (e.g. –250
m3/h)
• Upper range value output parameter (→  22) = not equal to zero flow (e.g. +750
m3/h)
• Calculated current value = 8 mA at zero flow

Endress+Hauser

21

"Guidance" menu

Proline Promag 10 Modbus RS485

I [mA]
20
84Q-250 0+750

11A0013757

Q
I
1

Flow
Current
Measuring range is exceeded or undershot

The linear operational range of the measuring device is defined by the values entered for
the Lower range value output parameter (→  21) and Upper range value output
parameter (→  22), and by the selected current range.
Example: Measuring mode with the "Forward/Reverse flow" option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

Q
I
1
2
3

Flow
Current
Value assigned to 0/4 mA current
Forward flow
Reverse flow

The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the Lower range value output parameter (→  21)
and Upper range value output parameter (→  22) must have the same algebraic sign.
The value for the Upper range value output parameter (→  22) (e.g. reverse flow)corresponds to the mirrored value for the Upper range value output parameter(→  22) (e.g. forward flow).
Example: Measuring mode with the "Reverse flow compensation" option
If flow is characterized by severe fluctuations (e.g. when using reciprocating pumps), flow
components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.

Upper range value output

Navigation

 Guidance → Commissioning → Upp.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

22

Endress+Hauser

Proline Promag 10 Modbus RS485"Guidance" menu

Description

Enter upper range value for the measured value range.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter: Lower range value output parameter (→  21):

Damping current output

Navigation

 Guidance → Commissioning → Damp.curr.outp

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter time constant (PT1 element) to set the reaction time of the output signal tofluctuations in the measured value caused by process conditions.
Additional information:- The smaller the time constant the faster the output reacts to fluctuations in themeasured value.
- If the time constant is 0, damping is deactivated.

User entry

0.0 to 999.9 s

Fixed current

Navigation

 Guidance → Commissioning → Fixed current

Prerequisite

In the Current range output parameter in the Current output 1 submenu, the Fixedvalue option is selected.

Description

Enter the value for the "Fixed value" option.

User entry

3.59 to 21.5 mA

Failure behavior current output

Navigation

 Guidance → Commissioning → Fail.behav.out

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Endress+Hauser

23

"Guidance" menu

Proline Promag 10 Modbus RS485

Description

Specify how the output should behave in the event of a device alarm.

Selection• Min.
• Max.
• Last valid value• Actual value• Fixed value

Additional information

Selection• Min. option
The current output outputs the lower fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Max. option
The current output outputs the upper fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Last valid value option
The current output outputs the last valid value measured before the device alarm
occurred.
• Actual value option
The current output outputs the flow value currently measured. The device alarm isignored.
• Fixed value option
The current output outputs the value specified.
Additional information:
The value is specified in the "Failure current " parameter.

Failure current

Navigation

 Guidance → Commissioning → Fail. current

Prerequisite

In the Failure behavior current output parameter in the Current output 1 submenu, theFixed value option is selected.

Description

Enter the value for the "Fixed value" option in the "Failure behavior current output "parameter.

User entry

3.59 to 21.5 mA

24Endress+Hauser

Proline Promag 10 Modbus RS485

2.1.7"Guidance" menu

Display

Navigation

 Guidance → Commissioning

Value 1 display

Navigation

 Guidance → Commissioning → Value 1 display

Description

Select the measured value that is displayed first on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 2 display

Navigation

 Guidance → Commissioning → Value 2 display

Description

Select the measured value that is shown second on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection*• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Visibility depends on order options or device settings

Endress+Hauser

25

"Guidance" menu

Proline Promag 10 Modbus RS485

Value 3 display

Navigation

 Guidance → Commissioning → Value 3 display

Description

Select the measured value that is shown third on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 4 display

Navigation

 Guidance → Commissioning → Value 4 display

Description

Select the measured value that is shown fourth on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Display damping

Navigation

 Guidance → Commissioning → Display damping

Description

Enter time constant (PT1 element) to set reaction time of the display to fluctuations in themeasured value.
Additional information:- The smaller the time constant the faster the display reacts to fluctuations in themeasured value.
- If the time constant is set to 0, damping is deactivated.

*

26

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485

User entry"Guidance" menu

0.0 to 999.9 s

2.1.8

Date/time

Navigation

 Guidance → Commissioning

Time format



Navigation

 Guidance → Commissioning → Time format

Description

Select time format.

Selection• 24 h• 12 h AM/PMTime zoneNavigation

 Guidance → Commissioning → Time zone

Description

Select the time zone. Every time the time zone is changed, a logbook entry is created.

Endress+Hauser

27

"Guidance" menu

Selection

Proline Promag 10 Modbus RS485

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+14:00

Set date/timeNavigationDescription

Set the date and local time. Every time the date or time is changed, a logbook entry iscreated.

User entry

Date and time

28

Guidance → Commissioning → Set date/time

Endress+Hauser

Proline Promag 10 Modbus RS485

3"Diagnostics" menu

"Diagnostics" menu

Troubleshooting and preventive maintenance – settings for device behavior during process
and device events as well as assistance and measures for diagnostic purposes.
Navigation

 Diagnostics

Diagnostics

Endress+Hauser‣ Active diagnostics→  30‣ Diagnostic list→  32‣ Simulation→  35‣ Heartbeat Technology→  38‣ Diagnostic settings→  3929

"Diagnostics" menu

Proline Promag 10 Modbus RS485

3.1"Actual diagnostics" submenu

Navigation

 Diagnostics → Active diagnos.

‣ Active diagnostics
Actual diagnostics→  30Timestamp→  30Previous diagnostics→  30Timestamp→  31

Operating time from restart→  31

Operating time→  31

Actual diagnostics
Navigation

 Diagnostics → Active diagnos. → Actual diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the currently active diagnostic message.
If there is more than one pending diagnostic event, the message for the diagnostic eventwith the highest priority is displayed.

User interface

Positive integer

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp for the currently active diagnostic message.

User interface

Days (d), hours (h), minutes (m), seconds (s)Previous diagnostics
Navigation

 Diagnostics → Active diagnos. → Prev.diagnostics

Prerequisite

At least two diagnostic events have already occurred.

30

Endress+Hauser

Proline Promag 10 Modbus RS485"Diagnostics" menu

Description

Displays the diagnostic message for the last diagnostic event that has ended.

User interface

Positive integer

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp of the diagnostic message generated for the last diagnostic eventthat has ended.

User interface

Days (d), hours (h), minutes (m), seconds (s)Operating time from restart
Navigation

 Diagnostics → Active diagnos. → Time fr. restart

Description

Indicates how long the device has been in operation since the last time the device wasrestarted.

User interface

Days (d), hours (h), minutes (m), seconds (s)Operating time
Navigation

 Diagnostics → Active diagnos. → Operating time

Description

Indicates how long the device has been in operation.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

31

"Diagnostics" menu

Proline Promag 10 Modbus RS485

3.2"Diagnostic list" submenu

Navigation

Diagnostics → Diagnostic list

‣ Diagnostic list
Diagnostics 1→  32

Timestamp→  32

Diagnostics 2→  33

Timestamp→  33

Diagnostics 3→  33

Timestamp→  33

Diagnostics 4→  33

Timestamp→  34

Diagnostics 5→  34

Timestamp→  34

Diagnostics 1
Navigation

Diagnostics → Diagnostic list → Diagnostics 1

Description

Displays the currently active diagnostic message with the highest priority.

User interface

Positive integer

Timestamp
NavigationDescription

Displays the timestamp for the diagnostic message with the highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)32Diagnostics → Diagnostic list → Timestamp

Endress+Hauser

Proline Promag 10 Modbus RS485"Diagnostics" menu

Diagnostics 2
Navigation

Diagnostics → Diagnostic list → Diagnostics 2

Description

Displays the currently active diagnostic message with the second highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the second highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 3
Navigation

Diagnostics → Diagnostic list → Diagnostics 3

Description

Displays the currently active diagnostic message with the third highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the third highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 4
NavigationDescription

Displays the currently active diagnostic message with the fourth highest priority.

User interface

Positive integer

Endress+Hauser

Diagnostics → Diagnostic list → Diagnostics 4

33

"Diagnostics" menu

Proline Promag 10 Modbus RS485

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the fourth highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 5
Navigation

Diagnostics → Diagnostic list → Diagnostics 5

Description

Displays the currently active diagnostic message with the fifth-highest priority.

User interface

Positive integer

Timestamp
NavigationDescription

Displays the timestamp for the diagnostic message with the fifth highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)34Diagnostics → Diagnostic list → Timestamp

Endress+Hauser

Proline Promag 10 Modbus RS485

3.3"Diagnostics" menu

"Simulation" submenu

Navigation

 Diagnostics → Simulation‣ Simulation
Assign simulation process variable→  35

Process variable value→  36

Current output 1 simulation→  36

Current output value→  36

Device alarm simulation→  37

Diagnostic event category→  37

Diagnostic event simulation→  37

Assign simulation process variable
Navigation

 Diagnostics → Simulation → Assign proc.var.

Description

Select a process variable for the simulation, thereby activating it.

Selection• Off• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *Additional information

DescriptionThe display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

*

Visibility depends on order options or device settings

Endress+Hauser

35

"Diagnostics" menu

Proline Promag 10 Modbus RS485

Process variable value

Navigation

 Diagnostics → Simulation → Proc. var. value

Description

Enter the simulation value for the selected process variable. Processing of measured values
downstream as well as the signal ouput follow this value. In this way, it is possibe to verify
whether the measuring device has been configured correctly.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User entry

Signed floating-point number

Current output 1 simulation



Navigation

 Diagnostics → Simulation → Curr.outp 1 sim.

Description

Switch simulation of the current output on or off.
Additional information:
When simulation is on, the current output signal is set to the value specified in the"Current output value" parameter.

Selection

• Off
• On

Additional information

Description
The display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

Current output value

Navigation

 Diagnostics → Simulation → Curr.outp val.

Description

Enter the current value for the simulation. In this manner, it is possible to verify thecurrent output is configured correctly and downstream processing units are functioning
properly.
Additional information:
The valid input range is determined by the "Current range output " parameter.

User entry

36

3.59 to 21.5 mA

Endress+Hauser

Proline Promag 10 Modbus RS485"Diagnostics" menu

Device alarm simulation



Navigation

 Diagnostics → Simulation → Dev. alarm sim.

Description

Switch the device alarm simulation on or off.
While simulation is in progress, the display alternates between the measured value and adiagnostic message of the Function Check (C) category.

Selection

• Off
• On

Diagnostic event category



Navigation

Diagnostics → Simulation → Event category

Description

Select the category of diagnostic events that should be available for selection in the"Diagnostic event simulation" parameter.

Selection• Sensor• Electronics• Configuration• ProcessDiagnostic event simulation



Navigation

 Diagnostics → Simulation → Diag. event sim.

Description

Select the diagnostic event to simulate.

Selection

Off

Endress+Hauser

37

"Diagnostics" menu

Proline Promag 10 Modbus RS485

3.4"Heartbeat Technology" submenu

The Heartbeat Technology submenu (→  38) is only available with the optional"Heartbeat Verification + Monitoring" application package.
• Order code for: Application package• Option: EB "Heartbeat Verification + Monitoring"
Detailed information and all descriptions of the device parameters of the application
package are available in the "Heartbeat Verification + Monitoring" Special
Documentation
Navigation

 Diagnostics → HBT‣ Heartbeat Technology

38

Endress+Hauser

Proline Promag 10 Modbus RS485

3.5"Diagnostics" menu

"Diagnostic settings" submenu

Navigation

 Diagnostics → Diag. settings‣ Diagnostic settings

3.5.1‣ Properties→  39‣ Diagnostic configuration→  39"Properties" submenu

Navigation

 Diagnostics → Diag. settings → Properties‣ Properties
Alarm delay→  39Alarm delay

Navigation

 Diagnostics → Diag. settings → Properties → Alarm delay

Description

Enter a duration for the alarm delay. When a diagnostic event of the "Alarm" category
occurs, the diagnostic message is not generated until the delay has elapsed.

User entry

0 to 60 s

3.5.2
Navigation"Diagnostic configuration" submenu
 Diagnostics → Diag. settings → Diag. config.

‣ Diagnostic configuration

Endress+Hauser‣ Sensor→  40‣ Electronics→  40‣ Configuration→  43‣ Process→  4339

"Diagnostics" menu

Proline Promag 10 Modbus RS485"Sensor" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Sensor‣ Sensor
Assign behavior of diagnostic no. 043→  40Assign behavior of diagnostic no. 043



Navigation

 Diagnostics → Diag. settings → Diag. config. → Sensor → Diagnostic no. 043

Description

Select behavior for diagnostic event "043 Sensor short circuit detected".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

"Electronics" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics‣ Electronics

40

Assign behavior of diagnostic no. 230→  41

Assign behavior of diagnostic no. 231→  41

Assign behavior of diagnostic no. 302→  41

Endress+Hauser

Proline Promag 10 Modbus RS485"Diagnostics" menu

Assign behavior of diagnostic no. 376→  41

Assign behavior of diagnostic no. 377→  42

Assign behavior of diagnostic no. 230
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 230

Description

Select behavior for diagnostic event "230 Date/time incorrect".

Selection• Alarm• Warning• Logbook entry only

Assign behavior of diagnostic no. 231
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 231

Description

Select behavior for diagnostic event "231 Date/time not available".

Selection• Alarm• Warning• Logbook entry only

Assign behavior of diagnostic no. 302
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 302

Description

Select behavior for diagnostic event "302 Device verification active".

Selection• Off• Warning• Logbook entry only

Assign behavior of diagnostic no. 376
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 376

Description

Select behavior for diagnostic event "376 Main electronics faulty".

Endress+Hauser









41

"Diagnostics" menu

Proline Promag 10 Modbus RS485

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 377

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 377

Description

Select behavior for diagnostic event "377 Sensor electronics (ISEM) faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

42

Endress+Hauser

Proline Promag 10 Modbus RS485"Diagnostics" menu

"Configuration" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration‣ Configuration
Assign behavior of diagnostic no. 441→  43

Assign behavior of diagnostic no. 441

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Diagnostic no. 441

Description

Select behavior for diagnostic event "441 Current output faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

"Process" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process‣ ProcessEndress+Hauser

Assign behavior of diagnostic no. 832→  44

Assign behavior of diagnostic no. 833→  44

Assign behavior of diagnostic no. 834→  45Assign behavior of diagnostic no. 835→  4543

"Diagnostics" menu

Proline Promag 10 Modbus RS485

Assign behavior of diagnostic no. 842→  46

Assign behavior of diagnostic no. 937→  46

Assign behavior of diagnostic no. 938→  47

Assign behavior of diagnostic no. 961→  47

Assign behavior of diagnostic no. 962→  48Assign behavior of diagnostic no. 832

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 832

Description

Select behavior for diagnostic event "832 Electronics temperature too high".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 833

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 833

Description

Select behavior for diagnostic event "833 Electronics temperature too low".

Selection• Off• Alarm• Warning• Logbook entry only

44

Endress+Hauser

Proline Promag 10 Modbus RS485

Additional information"Diagnostics" menu

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 834
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 834

Description

Select event category (status signal) for diagnostic event "834 Process temperature toohigh".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 835
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 835

Description

Select behavior for diagnostic event "835 Process temperature too low".

Selection• Off• Alarm• Warning• Logbook entry only

Endress+Hauser45

"Diagnostics" menu

Additional information

Proline Promag 10 Modbus RS485

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 842

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 842

Description

Select behavior for diagnostic event "842 Process value above limit".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 937

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 937

Description

Select behavior for diagnostic event "937 Sensor symmetry".

Selection• Off• Alarm• Warning• Logbook entry only46

Endress+Hauser

Proline Promag 10 Modbus RS485

Additional information"Diagnostics" menu

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 938
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 938

Description

Select behavior for diagnostic event "938 EMC interference".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 961
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 961

Description

Select behavior for diagnostic event "961 Electrode potential out of specification".

Selection• Off• Alarm• Warning• Logbook entry only

Endress+Hauser

47

"Diagnostics" menu

Additional information

Proline Promag 10 Modbus RS485

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 962

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 962

Description

Select behavior for diagnostic event "962 Pipe empty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

48

Endress+Hauser

Proline Promag 10 Modbus RS485

4"Application" menu

"Application" menu

Targeted optimization to the application – comprehensive device settings from sensor
technology to system integration for optimum application adaptation.
Navigation

 Application

Application

4.1‣ Measured values→  49‣ System units→  53‣ Totalizers→  58‣ Sensor→  63‣ Current output 1→  74‣ Communication→  85"Measured values" submenu

Navigation

 Application → Measured values‣ Measured values
Mass flow→  49Volume flow

→  50Conductivity→  50

Corrected conductivity→  50Temperature→  51‣ Totalizer→  51Mass flow
Navigation

 Application → Measured values → Mass flow

Description

Displays the mass flow currently calculated.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Endress+Hauser

49

"Application" menu

User interface

Proline Promag 10 Modbus RS485

Signed floating-point number

Volume flow
Navigation

 Application → Measured values → Volume flow

Description

Displays the volume flow currently measured.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Signed floating-point number

Conductivity
Navigation

 Application → Measured values → Conductivity

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the conductivity currently measured.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Positive floating-point number

Corrected conductivity
Navigation

 Application → Measured values → CorrConductivity

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the currently measured conductivity compensated for temperature.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

50

Positive floating-point number

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

Temperature
Navigation

 Application → Measured values → Temperature

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Displays the medium temperature currently measured.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Positive floating-point number

4.1.1"Totalizer" submenu

Navigation

 Application → Measured values → Totalizer‣ Totalizer
Totalizer value 1 to n

→  51

Totalizer overflow 1 to n

→  52

Totalizer value 1 to n

Navigation

 Application → Measured values → Totalizer → Totalizer val. 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Displays the current totalizer counter.
Additional information:
Since the operating tool cannot display figures with more than 7 digits, the current counter
above this range equals the sum of the totalizer counter plus the overflow displayed for the"Totalizer overflow " parameter.
Example for how to calculate the current totalizer counter when the value exceeds the 7digit display range limit of the operating tool:- Value of "Totalizer value " parameter: 1,968,457 m³- Value of "Totalizer overflow " parameter: 1 × 107 m³ = 10,000,000 m³- Current totalizer reading: 11,968,457 m³
In the event of an error, the totalizer behaves as specified in the "Failure mode" parameter.

User interface

Endress+Hauser

Signed floating-point number

51

"Application" menu

Proline Promag 10 Modbus RS485

Totalizer overflow 1 to n

Navigation

 Application → Measured values → Totalizer → Tot. overflow 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Displays the current totalizer overflow.
Additional information:
If the current totalizer counter exceeds the operating tool's maximum numerical display
range of 7 digits, the amount above this range is expressed as an overflow. The current
totalizer counter therefore equals the sum of the overflow and the totalizer value displayedin the "Totalizer value " parameter.
Example of how to calculate the current totalizer counter when the value exceeds the 7digit display limit of the operating tool:- Value of "Totalizer value " parameter: 1,968,457 m³- Value of "Totalizer overflow " parameter: 1 × 10^7 m³ = 10,000,000 m³- Current totalizer reading: 11,968,457 m³

User interface

52–32 000.0 to 32 000.0

Endress+Hauser

Proline Promag 10 Modbus RS485

4.2"Application" menu

"System units" submenu

Navigation

 Application → System units‣ System units
Volume flow unit

→  53

Volume unit

→  54Mass flow unit

→  55Mass unit→  55

Density unit

→  56

Temperature unit

→  56

Conductivity unit

→  56

Volume flow unitNavigation

 Application → System units → Volume flow unit

Description

Select volume flow unit.

Endress+Hauser

53

"Application" menu

Selection

Proline Promag 10 Modbus RS485

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dUS units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• MMft³/s• MMft³/min• MMft³/h• Mft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)Volume unitNavigation

 Application → System units → Volume unit

Description

Select volume unit.

54

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)Endress+Hauser

Proline Promag 10 Modbus RS485

Selection

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega"Application" menu

US units• af• ft³• Mft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;beer)• bbl (us;tank)

Mass flow unitNavigation

 Application → System units → Mass flow unit

Description

Select mass flow unit.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

Mass unitNavigation

 Application → System units → Mass unit

Description

Select mass unit.

Selection

SI units
•g
• kg
•t

Endress+Hauser

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

US units• oz• lb• STon55

"Application" menu

Proline Promag 10 Modbus RS485

Density unitNavigation

 Application → System units → Density unit

Description

Select density unit.

Selection

SI units• g/cm³• g/m³• kg/l• kg/dm³• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

US units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)Temperature unitNavigation

 Application → System units → Temperature unit

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Select temperature unit.

Selection

SI units• °C•K

US units• °F• °R

Conductivity unitNavigation

 Application → System units → Conductiv. unit

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

56

Select conductivity unit.

Endress+Hauser

Proline Promag 10 Modbus RS485

Selection

Endress+Hauser"Application" menu

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/m
• kS/m• MS/m57

"Application" menu

Proline Promag 10 Modbus RS485

4.3"Totalizers" submenu

Navigation

 Application → Totalizers‣ Totalizers

4.3.1‣ Totalizer handling→  58‣ Totalizer 1 to n

→  58"Totalizer handling" submenu

Navigation

 Application → Totalizers → Totalizer‣ Totalizer handlingReset all totalizers→  58Reset all totalizers
Navigation

 Application → Totalizers → Totalizer → Reset all tot.

Description

Reset all totalizers to "0" and restart the totaling process. All flow quantities thus far
totalized are thereby deleted.

Selection• Cancel• Reset + totalize

4.3.2"Totalizer 1 to n" submenu

Navigation

 Application → Totalizers → Totalizer 1 to n

‣ Totalizer 1 to n

58

Assign process variable→  59Unit totalizer 1 to n

→  59Totalizer operation mode→  60Control Totalizer 1 to n

→  60Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

Preset value 1 to n

→  61

Failure mode→  61

Assign process variable

Navigation

 Application → Totalizers → Totalizer 1 to n → Assign variable

Description

Select process variable for totalizer.
Additional information:
If the option selected is changed, the device resets the totalizer to "0".

Selection• Off• Volume flow
• Mass flow

Unit totalizer 1 to n

Navigation

 Application → Totalizers → Totalizer 1 to n → Unit totalizer 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select process variable totalizer unit.

Selection

SI units• g*• kg *• t**

US units• oz *• lb *• STon *Visibility depends on order options or device settingsorSI units• cm³ *
• dm³ *
• m³ *• ml *• l*• hl *• Ml Mega **Endress+Hauser

US units• af *• ft³ *• Mft³ *• fl oz (us) *• gal (us) *• kgal (us) *• Mgal (us) *• bbl (us;liq.) *• bbl (us;beer) *• bbl (us;oil) *• bbl (us;tank) *Imperial units• gal (imp) *• Mgal (imp) *• bbl (imp;beer) *• bbl (imp;oil) *Visibility depends on order options or device settings

59

"Application" menu

Proline Promag 10 Modbus RS485

or
Other units
None **

Additional information

Visibility depends on order options or device settings

Description
The unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  53).
Selection
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  13).

Totalizer operation modeNavigation

 Application → Totalizers → Totalizer 1 to n → Operation mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select totalizer calculation mode.

Selection• Net flow total• Forward flow total• Reverse flow total

Additional information

Selection• Net flow total option
The flow values in the forward and reverse flow directions are totalized and netted
against each other. Net flow is recorded in the flow direction.
• Forward flow total option
Only the flow in the forward flow direction is totalized.
• Reverse flow total option
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Control Totalizer 1 to n
Navigation

 Application → Totalizers → Totalizer 1 to n → Control Tot. 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Operate the totalizer.

60

Endress+Hauser

Proline Promag 10 Modbus RS485

Selection• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• HoldAdditional information

Selection"Application" menu

• Totalize option
The totalizer is started or continues running.
• Reset + hold option
The totaling process is stopped and the totalizer is reset to "0".
• Preset + hold option
The totaling process is stopped and the totalizer is set to the start value specified in the"Preset value " parameter.
• Reset + totalize option
The totalizer is reset to "0" and the totaling process is restarted.
• Hold option
Totalizing is stopped.

Preset value 1 to n
Navigation

 Application → Totalizers → Totalizer 1 to n → Preset value 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify start value for totalizer.

User entry

Signed floating-point number

Additional information

Description
The unit of the selected process variable is specified for the totalizer in the Unit totalizer
parameter (→  13).
ExampleThis configuration is suitable for applications such as iterative filling processes with a fixed
batch quantity.

Failure modeNavigation

 Application → Totalizers → Totalizer 1 to n → Failure mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify how the totalizer should behave in the event of a device alarm.
Additional information:
The failsafe mode that applies to any other totalizers or outputs is specified separately inother parameters and is not impacted by this setting.

Endress+Hauser

61

"Application" menu

Proline Promag 10 Modbus RS485

Selection• Stop• Actual value• Last valid value

Additional information

Selection• Stop option
The totalizer is stopped in the event of a device alarm.
• Actual value option
The totalizer continues to totalize based on the current value measured; the device alarmis ignored.
• Last valid value option
The totalizer continues to totalize based on the last valid value measured before thedevice alarm occurred.

62

Endress+Hauser

Proline Promag 10 Modbus RS485

4.4
Navigation"Application" menu

"Sensor" submenu
 Application → Sensor‣ Sensor

4.4.1
Navigation‣ Process parameters→  63‣ Low flow cut off→  66‣ Empty pipe detection→  67‣ Sensor adjustment→  69‣ Calibration→  70‣ Electrode cleaning cycle→  72"Process parameters" submenu
 Application → Sensor → Process param.

‣ Process parameters

Endress+Hauser

Flow damping→  64

Flow override→  64

Conductivity measurement→  64

Conductivity temperature coefficient→  65

Conductivity damping time→  65Temperature damping time→  66

Fixed density→  6663

"Application" menu

Proline Promag 10 Modbus RS485

Flow damping

Navigation

 Application → Sensor → Process param. → Flow damping

Description

Enter value for damping of the flow measured value in order to reduce the variability of
the flow measured value when exposed to interference.
Additional information:
The depth of the flow filter is determined by this setting. As the filter depth increases, sodoes the reaction time of the device.
- Value = 0: No damping. Damping of 0 is not recommended, as the measuring signal isthen so noisy that it is almost impossible to perform a measurement.
- Value > 0: Damping increases
Optimal damping depends on the measuring period.
Damping impacts the following measuring device variables:- Outputs- Low flow cut off- Totalizers

User entry

0 to 15

Flow override

Navigation

 Application → Sensor → Process param. → Flow override

Description

Stops the measuring process. Can be used for example when cleaning the pipeline.

Selection

• Off
• On

Additional information

Selection"On" option
Activates flow override. The diagnostic message "453 Flow override active" is generated.
Additional information:
Output values:- Temperature: Measurement continues- Totalizers 1 to 3: No longer totalize

Conductivity measurement

Navigation

 Application → Sensor → Process param. → Conduct. measur.

Prerequisite

Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option", option
CX (conductivity measurement)

64

Endress+Hauser

Proline Promag 10 Modbus RS485

Description"Application" menu

Switch conductivity measurement on or off.
Additional information:
To be able to measure conductivity, the medium must have a minimum conductivity of 5
μS/cm.

Selection

• Off
• On

Conductivity temperature coefficientNavigation

 Application → Sensor → Process param. → Cond. temp.coeff

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Enter the temperature coefficient for conductivity.

User entry

Signed floating-point number

Conductivity damping timeNavigation

Application → Sensor → Process param. → ConductDampTime

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Enter time constant for conductivity damping (PT1 element):- Value = 0: No damping- Value > 0: Damping increases
Additional information:
Damping is implemented by means of a proportional transmission behavior with firstorder delay (PT1 element).

User entry

Endress+Hauser

0 to 999.9 s

65"Application" menu

Proline Promag 10 Modbus RS485

Temperature damping timeNavigation

Application → Sensor → Process param. → TempDampingTime

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Enter time constant for damping the temperature value.

User entry

0 to 999.9 s

Fixed density

Navigation

 Application → Sensor → Process param. → Fixed density

Description

Enter a fixed value for the density.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User entry

Positive floating-point number

4.4.2"Low flow cut off" submenu

Navigation

 Application → Sensor → Low flow cut off‣ Low flow cut off
Low flow cut off→  66

On value low flow cutoff→  67

Off value low flow cutoff→  67

Low flow cut offNavigation

 Application → Sensor → Low flow cut off → Low flow cut off

Description

Select process variable for low flow cut off to activate low flow cut off.

Selection• Off• Volume flow
• Mass flow

66

Endress+Hauser

Proline Promag 10 Modbus RS485

Additional information"Application" menu

Description
Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn-value entered
Off-value entered

On value low flow cutoff

Navigation

 Application → Sensor → Low flow cut off → On value

Description

Enter on value to switch on low flow cut off.
Value = 0: No low flow cut off
Value > 0: Low flow cut off is activatedUser entry

Positive floating-point number

Off value low flow cutoff

Navigation

 Application → Sensor → Low flow cut off → Off value

Description

Enter off value to switch off low flow cut off. The off value is entered as a positive
hysteresis with respect to the on value.

User entry

0 to 100.0 %4.4.3
Navigation"Partially filled pipe detection" submenu
 Application → Sensor → Partial pipe det‣ Empty pipe detection
Empty pipe detection

Endress+Hauser→  6867

"Application" menu

Proline Promag 10 Modbus RS485

New adjustment→  68Empty pipe adjust value→  68Full pipe adjust value→  69Measured value EPD→  69Empty pipe detection



Navigation

 Application → Sensor → Empty pipe det. → Empty pipe det.

Description

Switch empty pipe detection on or off. Switch on empty pipe detection to detect a partiallyfilled or empty measuring tube.

Selection

• Off
• On

New adjustment

Navigation

 Application → Sensor → Empty pipe det. → New adjustment

Description

Select empty pipe or full pipe adjustment to perform a new adjustment. To adjust empty
pipe detection, perform the empty pipe adjustment first and then the full pipe adjustment.
Additional information:
The measuring device is pre-adjusted at production using water (approx. 300 μS/cm). For
liquids that deviate from this conductivity, a new empty pipe and full pipe adjustment mustbe performed on site.

Selection• Cancel• Empty pipe adjust• Full pipe adjust

Empty pipe adjust value

Navigation

 Application → Sensor → Empty pipe det. → Empty pipe value

Description

Displays adjustment value when the measuring tube is empty.
NOTE
Users logged on in the Service role have write access!

User interface

68

Positive floating-point number

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

Full pipe adjust value

Navigation

 Application → Sensor → Empty pipe det. → Full pipe value

Description

Displays adjustment value when the measuring tube is full.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

Measured value EPD
Navigation

 Application → Sensor → Empty pipe det. → Meas. value EPD

Description

Displays the value currently measured for empty pipe detection.

User interface

Positive floating-point number

4.4.4"Sensor adjustment" submenu

Navigation

 Application → Sensor → Sensor adjustm.

‣ Sensor adjustment
Installation direction→  69Integration time→  70Measuring period→  70Installation direction



Navigation

 Application → Sensor → Sensor adjustm. → Install. direct.

Description

Select sign of flow direction

Selection• Forward flow
• Reverse flow

Endress+Hauser

69

"Application" menu

Proline Promag 10 Modbus RS485

Integration timeNavigation

 Application → Sensor → Sensor adjustm. → Integration time

Description

Displays the duration of an integration cycle.
NOTE
Users logged on in the Service role have write access!

User interface

1 to 65 ms

Measuring period

Navigation

 Application → Sensor → Sensor adjustm. → Measuring period

Description

Displays the duration of a full measuring period.
Additional information:
The measuring period is the time span during which the excitation of the magnetic field
takes place and a measuring point is created.
NOTE
Users logged on in the Service role have write access!

User interface

0 to 1 000 ms

4.4.5"Calibration" submenu

Navigation

 Application → Sensor → Calibration‣ Calibration
Nominal diameter→  70Calibration factor→  71

Zero point→  71

Conductivity calibration factor→  71

Nominal diameter
Navigation

 Application → Sensor → Calibration → Nominal diameter

Description

Shows the nominal diameter of the sensor.

70

Endress+Hauser

Proline Promag 10 Modbus RS485

User interface"Application" menu

Character string comprising numbers, letters and special characters

Calibration factor
Navigation

 Application → Sensor → Calibration → Cal. factor

Description

Displays the current calibration factor for the flow rate measuring sensor.
Additional information:
The factory setting for the calibration factor can be found on the sensor's nameplate.

User interface

Positive floating-point number

Zero pointNavigation

 Application → Sensor → Calibration → Zero point

Description

Displays the zero point correction value for the sensor.
NOTE
Users logged on in the Service role have write access!

User interface

Signed floating-point number

Conductivity calibration factor



Navigation

 Application → Sensor → Calibration → Cond. cal. fact.

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  64).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays calibration factor for conductivity measurement.
NOTE
Users logged on in the Service role have write access!

User interface

Endress+Hauser

0.01 to 10 000

71

"Application" menu

Proline Promag 10 Modbus RS485

4.4.6"Electrode cleaning cycle" submenu

Navigation

 Application → Sensor → Elec. clean cycl

‣ Electrode cleaning cycle
Electrode cleaning cycle→  72

ECC duration→  72

ECC recovery time→  72

ECC interval→  73

ECC polarity→  73

Electrode cleaning cycleNavigation

 Application → Sensor → Elec. clean cycl → Elec. clean cycl

Description

Switch electrode cleaning on or off.

Selection

• Off
• On

ECC duration



Navigation

 Application → Sensor → Elec. clean cycl → ECC duration

Description

Enter the duration of the cleaning cycle.

User entry

0.01 to 30 s

ECC recovery timeNavigation

 Application → Sensor → Elec. clean cycl → ECC recov. time

Description

Specify a timespan for recovery after a cleaning cycle has completed to prevent
interference with the signal outputs. The output signal values will be frozen for theduration of the recovery.

User entry

1 to 600 s

72

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

ECC interval

Navigation

 Application → Sensor → Elec. clean cycl → ECC interval

Description

Specify the duration of the interval between one cleaning cycle and the next.

User entry

0.5 to 168 h

ECC polarity
Navigation

 Application → Sensor → Elec. clean cycl → ECC polarity

Description

Displays the setting for the electrode cleaning polarity.
Additional information:
The polarity depends on the material of the electrodes:- Platinum: Negative- Tantalum, Alloy C22, stainless steel: Positive

User interface

Endress+Hauser• Positive• Negative

73

"Application" menu

Proline Promag 10 Modbus RS485

4.5"Current output" submenu

Navigation

Application → Curr.output 1‣ Current output 1
Process variable current output→  74

Measuring mode current output→  75Current range output→  79Fixed current→  80Lower range value output→  80Upper range value output→  82

Damping current output→  82

Failure behavior current output→  83

Failure current→  83

Output current 1→  84

Process variable current output
Navigation

 Application → Curr.output 1 → Proc.var. outp

Description

Select process variable for current output

Selection• Off *• Mass flow
• Conductivity *• Corrected conductivity *• Volume flow
• Temperature *• Noise *• Coil current shot time *

*

74



Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

Measuring mode current output

Navigation

 Application → Curr.output 1 → Meas.mode outp

Prerequisite

A process variable is selected in the Process variable current output parameter(→  20).

Description

Select the measuring mode for the output.

Selection• Forward flow
• Forward/Reverse flow *• Reverse flow compensation*

Visibility depends on order options or device settings

Endress+Hauser

75

"Application" menu

Additional information

Proline Promag 10 Modbus RS485

Selection• Forward flow option
The current output signal is proportional to the measured value for the process variableassigned.
Additional information:- The lower limit value ("Lower range value output " parameter) and the upper limit value("Upper range value output " parameter) of the measured value range do not have to havethe same algebraic sign, i.e. the lower limit value can be negative and the upper limit
value positive.
- If the measured value lies outside the scaled measured value range, diagnostic message"441 Current output faulty" is generated.
• Forward/Reverse flow option
The current output outputs the absolute value for the assigned process variable(reflection across the lower limit value of the measured value range).
Additional information:- The lower limit value ("Lower range value output " parameter) and the upper limit value("Upper range value output " parameter) of the measured value range must have thesame algebraic sign.
- If the absolute value exceeds the upper limit value of the measured value range,
diagnostic message "441 Current output faulty" is generated.
- This setting is generally only used for flow-related process variables.
• Reverse flow compensation option
The current output signal is proportional to the measured value for the process variableassigned.
Additional information:- The lower limit value ("Lower range value output " parameter) and the upper limit value("Upper range value output " parameter) of the measured value range do not have to havethe same algebraic sign, i.e. the lower limit value can be negative and the upper limit
value positive.
- Reverse flow (a measured value below the lower limit value of the measured value
range) is stored in a buffer and processed after a maximum delay of 60 s with the next
forward flow.
- When the flow exceeds the maximum value or the reverse flow stored in the buffer
cannot be processed within approx. 60 s, diagnostic message "441 Current output faulty"
is generated.
- This option is used e.g. to compensate intermittent reverse flow, which may occur inconnection with positive displacement pumps as a result of wear and tear or highviscosity.
- There is no flow damping with this setting.
1. Examples of the behavior of the current output
Defined measuring range: lower range value and upper range value with the same
algebraic sign.
I [mA]
20

4
0

1

2

Q
A0028084

I
Q
1
2

76

Current
Flow
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menuWith the following flow response:Qt
A0028091

Q
t

Flow
Time

With the Forward flow option
The current output signal is proportional to the process variable assigned. The flow
components outside the scaled measuring range are not taken into account for signal
output:.
I [mA]

t
A0028092

It

Current
Time

With the Forward/Reverse flow option
The current output signal is independent of the direction of flow.
I [mA]

t
A0028093

It

Current
Time

With the Reverse flow compensation option
Flow components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.

Endress+Hauser

77

"Application" menu

Proline Promag 10 Modbus RS485

I [mA]At

S
S=A

A0028094

It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

2. Examples of the behavior of the current output
Defined measuring range: lower range value and upper range value with different
algebraic signs.
I [mA]
20
4
0

Q

0
1

2
A0028095

I
Q
1
2

Current
Flow
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With flow a (—) outside, b (- -) inside the measuring range:
Q
1
+

t

0
2

-

a=
b=
A0028098

Q
t
a
b

Flow
Time
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With the Forward flow option• a (—): The flow components outside the scaled measuring range cannot be taken into
account for signal output. The diagnostic message S441 Current output 1 to ndiagnostic message is displayed.
• b (- -): The current output signal is proportional to the process variable assigned.

78

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

I [mA]ta=b=
A0028100It

Current
Time

With the Forward/Reverse flow option
This option cannot be selected here since the values for the 20 mA value parameter(→  22) and 20 mA value parameter (→  22) have different algebraic signs.
With the Reverse flow compensation option
Flow components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.
I [mA]At

S
S=A

A0028101It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

Current range output

Navigation

 Application → Curr.output 1 → Curr.range out

Description

Select the current range for the measured value output and the upper and lower fault
condition signal level.
Additional information:- The measured value range is specified in the "Lower range value output " parameter andthe "Upper range value output " parameter.
- If the measured value lies outside the scaled measured value range, diagnostic message"441 Current output faulty" is generated.
- In the event of a device alarm, the current output adopts the behavior specified in the"Failure behavior current output " parameter.

Selection

Endress+Hauser• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)• Fixed value

79

"Application" menu

Additional information

Proline Promag 10 Modbus RS485

Selection• 4...20 mA NE (3.8...20.5 mA) option
Select this option to set the current range in accordance with NAMUR recommendation
NE43.
• Fixed value option
Select this option to set the current output to a current value instead of a range.
The current value is defined in the Fixed current parameter (→  23).
The graphic shows the relationship between the current range for the output of theprocess value and the lower and upper alarm levels:
2

1

3

I [mA]
A0034351

1
2
3

Current range for process value
Lower level for signal on alarm
Upper level for signal on alarm

Selection (current range for process value) Lower level for signal on
alarm

Upper level for signal on
alarm

4...20 mA NE (3.8...20.5 mA)> 21.5 mA< 3.6 mA

4...20 mA US (3.9...20.8 mA)
4...20 mA (4... 20.5 mA)

Fixed current

Navigation

 Application → Curr.output 1 → Fixed current

Prerequisite

In the Current range output parameter in the Current output 1 submenu, the Fixedvalue option is selected.

Description

Enter the value for the "Fixed value" option.

User entry

3.59 to 21.5 mA

Lower range value output

Navigation

 Application → Curr.output 1 → Low.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

80

Endress+Hauser

Proline Promag 10 Modbus RS485

Description"Application" menuEnter lower range value for the measured value range.
Additional information:- Depending on the setting selected for the "Measuring mode current output " parameter,
the value specified for this parameter and the "Upper range value output " parameter musthave the same algebraic sign or not.
- As a rule, the lower range value is scaled to be lower than the upper range value. As aresult, the behavior of the current output is proportional to the process variable assigned.
If the lower range value is scaled to be higher than the upper range value, then the
behavior of the current output will be inversely proportional to the process variableassigend.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter (→  75).
Example: Measuring mode with "Forward flow" option• Lower range value output parameter (→  21) = not equal to zero flow (e.g. –250
m3/h)
• Upper range value output parameter (→  22) = not equal to zero flow (e.g. +750
m3/h)
• Calculated current value = 8 mA at zero flow
I [mA]
20
84Q-250 01+750
1A0013757

Q
I
1

Flow
Current
Measuring range is exceeded or undershot

The linear operational range of the measuring device is defined by the values entered for
the Lower range value output parameter (→  21) and Upper range value output
parameter (→  22), and by the selected current range.
Example: Measuring mode with the "Forward/Reverse flow" option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

Q
I
1
2
3

Endress+Hauser

Flow
Current
Value assigned to 0/4 mA current
Forward flow
Reverse flow

81

"Application" menu

Proline Promag 10 Modbus RS485

The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the Lower range value output parameter (→  21)
and Upper range value output parameter (→  22) must have the same algebraic sign.
The value for the Upper range value output parameter (→  22) (e.g. reverse flow)corresponds to the mirrored value for the Upper range value output parameter (→  22)(e.g. forward flow).
Example: Measuring mode with the "Reverse flow compensation" option
If flow is characterized by severe fluctuations (e.g. when using reciprocating pumps), flow
components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.

Upper range value output

Navigation

 Application → Curr.output 1 → Upp.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter upper range value for the measured value range.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter: Lower range value output parameter (→  21):

Damping current output

Navigation

 Application → Curr.output 1 → Damp.curr.outp

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter time constant (PT1 element) to set the reaction time of the output signal tofluctuations in the measured value caused by process conditions.
Additional information:- The smaller the time constant the faster the output reacts to fluctuations in themeasured value.
- If the time constant is 0, damping is deactivated.

User entry

82

0.0 to 999.9 s

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

Failure behavior current output

Navigation

 Application → Curr.output 1 → Fail.behav.out

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Specify how the output should behave in the event of a device alarm.

Selection• Min.
• Max.
• Last valid value• Actual value• Fixed value

Additional information

Selection• Min. option
The current output outputs the lower fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Max. option
The current output outputs the upper fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Last valid value option
The current output outputs the last valid value measured before the device alarm
occurred.
• Actual value option
The current output outputs the flow value currently measured. The device alarm isignored.
• Fixed value option
The current output outputs the value specified.
Additional information:
The value is specified in the "Failure current " parameter.

Failure current

Navigation

 Application → Curr.output 1 → Fail. current

Prerequisite

In the Failure behavior current output parameter in the Current output 1 submenu, theFixed value option is selected.

Description

Enter the value for the "Fixed value" option in the "Failure behavior current output "parameter.

User entry

3.59 to 21.5 mA

Endress+Hauser

83

"Application" menu

Proline Promag 10 Modbus RS485

Output current
Navigation

 Application → Curr.output 1 → Output curr. 1

Description

Displays the current value currently calculated.

User interface

3.59 to 21.5 mA

84

Endress+Hauser

Proline Promag 10 Modbus RS485

4.6"Application" menu

" Communication" submenu

Navigation

 Application → Communication‣ Communication

4.6.1
Navigation‣ Modbus configuration→  85‣ Modbus data map→  88‣ Modbus information→  89"Modbus configuration" submenu
 Application → Communication → Modbus config.

‣ Modbus configuration
Bus address→  85

Baudrate→  86

Parity→  86

Byte order→  87

Telegram delay→  87

Failure mode→  87

Fieldbus writing access→  88Bus addressNavigation

 Application → Communication → Modbus config. → Bus address

Description

Enter device address.

User entry

1 to 247

Endress+Hauser

85

"Application" menu

Proline Promag 10 Modbus RS485

Baudrate

Navigation

 Application → Communication → Modbus config. → Baudrate

Description

Define data transfer speed.

Selection• 1200 BAUD
• 2400 BAUD
• 4800 BAUD
• 9600 BAUD
• 19200 BAUD
• 38400 BAUD
• 57600 BAUD
• 115200 BAUD

Parity

Navigation

 Application → Communication → Modbus config. → Parity

Description

Select parity bits.
Additional information:"ASCII" picklist option:- 0 = "Even" option- 1 = "Odd" option"RTU" picklist option:- 0 = "Even" option- 1 = "Odd" option- 2 = "No parity bit/1 stop bit" option- 3 = "No parity bit/2 stop bits" option

Selection

86

• Odd
• Even• None / 1 stop bit• None / 2 stop bits

Endress+Hauser

Proline Promag 10 Modbus RS485"Application" menu

Byte orderNavigation

 Application → Communication → Modbus config. → Byte order

Description

Select byte transmission sequence. The transmission sequence must be coordinated with
the Modbus master.
Additional information:
The byte sequence is not standardized by the Modbus protocol. However, if the hostsystem and the measuring device do not use the same byte sequence, correct dataexchange is not possible.
Changing the byte sequence in the host system often requires extensive knowledge andsignificant programming effort. This parameter can be used to keep the standard settingsof the host system, while adjusting the byte sequence on the measuring device by meansof trial and error. If it is not possible to achieve correct data exchange in this manner, thenit is the host system's settings for the byte sequence that must be modified.

Selection• 0-1-2-3• 3-2-1-0• 1-0-3-2• 2-3-0-1

Telegram delay

Navigation

 Application → Communication → Modbus config. → Telegram delay

Description

Enter a delay time after which the measuring device replies to the request telegram of the
Modbus master. This is to enable communication with slow Modbus RS485 masters.

User entry

0 to 100 ms

Failure modeNavigation

 Application → Communication → Modbus config. → Failure mode

Description

Select the preferred Modbus communication output mode for the measured value, when adiagnostic event of the category specified in the "Assign diagnostic behavior" parameteroccurs.

Selection• NaN value• Last valid value

Additional information

Selection• NaN value option
The device outputs the NaN value ("Not a number" value).
• Last valid value option
The device displays the last valid measured value before the issue occurred.

Endress+Hauser

87

"Application" menu

Proline Promag 10 Modbus RS485

Fieldbus writing access
Navigation

 Application → Communication → Modbus config. → Fieldb.writ.acc.

Description

Restrict access to the measuring device via Fieldbus.
Additional information:
Once read and/or write protection has been enabled, this parameter can only be accessed
and reset via local operation. Access via an operating tool is no longer possible.
NOTE
The cyclical transmission of measured values to the higher-order system is not impacted
by these restrictions and always guaranteed!

Selection• Read + write• Read only

4.6.2

"Modbus data map" submenu

Navigation

 Application → Communication → Modbus data map‣ Modbus data map
Scan list register 0 to 15→  88

Scan list register 0 to 15

Navigation

 Application → Communication → Modbus data map → Scan list reg.0 to 15

Description

Enter the scan list register.
Additional information:
By entering the register address (1-based), it is possible to group up to 16 device
parameters, which thereby are assigned to the scan list registers 0 to 15. The data
corresponding to the device parameters assigned is read out via the register addresses
5051 to 5081.

User entry

88

0 to 65 535

Endress+Hauser

Proline Promag 10 Modbus RS485

4.6.3"Application" menu

"Modbus information" submenu

Navigation

 Application → Communication → Modbus info

‣ Modbus information
Device ID→  89Device revision→  89Device ID
Navigation

 Application → Communication → Modbus info → Device ID

Description

Displays the device ID to identify the measuring device.

User interface

0 to 65 535

Device revision
Navigation

 Application → Communication → Modbus info → Device revision

Description

Displays device revision.

User interface

0 to 65 535

Endress+Hauser

89

"System" menu

Proline Promag 10 Modbus RS485

5"System" menu

Overall device management and security settings – management of system settings andadaption to operational requirements.
Navigation

 System

System90‣ Device management→  91‣ User management→  93‣ Connectivity→  96‣ Date/time→  97‣ Information→  99‣ Display→  104‣ Software configuration→  108

Endress+Hauser

Proline Promag 10 Modbus RS485

5.1"System" menu

"Device management" submenu

Navigation

 System → Device manag.

‣ Device management
Device tag→  91

Locking status→  91

Configuration counter→  92

Device reset→  92

Device tagNavigation

 System → Device manag. → Device tag

Description

Enter a unique designation for the measuring point to be able to easily identify it within
the plant.

User entry

Character string comprising numbers, letters and special characters (32)

Locking status
Navigation

 System → Device manag. → Locking status

Description

Indicates the write protection with the highest priority that is currently active.

User interface• Hardware locked• Temporarily locked

Additional information

User interface• Hardware locked option
The DIP switch for the hardware lock is enabled. As a result write access to theparameters is locked.
• Temporarily locked option
Due to internal procedures that are currently in progress (e.g. data upload/download,
reset, etc.), write access to the parameters is temporarily locked.The parameters can be
modified again, once the internal procedures are complete.

Endress+Hauser

91

"System" menu

Proline Promag 10 Modbus RS485

Configuration counter
Navigation

 System → Device manag. → Config. counter

Description

Displays the counter for changes to the device parameters.
Additional information:- If the value for a static parameter is changed when optimizing or configuring theparameter, the counter is incremented by 1. This is to enable tracking different parameterversions.
- When multiple parameters are changed simultaneously, e.g. when loading parameters
into the device from an external source such as FieldCare, the counter may display a higher
value. The counter cannot be reset, nor is it reset to a default value on performing a device
reset.
- Once the counter has reached the value 65535, it restarts at 0.

User interface

0 to 65 535

Device reset

Navigation

 System → Device manag. → Device reset

Description

Reset the device configuration - either entirely or in part - to a defined state.

Selection• Cancel• To delivery settings• Restart device• Restore S-DAT backup *• Create T-DAT backup• Restore T-DAT backup *Additional information

Selection• To delivery settings option
Every parameter for which a customer-specific default setting was ordered is reset to thecustomer-specific value. All other parameters are reset to the factory setting.
• Restart device option
The restart resets every parameter with data stored in volatile memory (RAM) to thefactory setting (e.g. measured value data). The device configuration remains unchanged.
• Restore S-DAT backup option
Restore the data that is saved on the S-DAT. The data record is restored from theelectronics memory to the S-DAT.
• Create T-DAT backup option
Create T-DAT backup.

*

92

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485

5.2"System" menu

"User management" submenu

Navigation

 System → User manag.

‣ User management
User role→  93Enter access code→  94

Reset Maintenance code→  94‣ Define Maintenance code→  95User role
Navigation

 System → User manag. → User role

Description

Displays the role the user is currently logged on in. The role determines the user's accessrights for the parameters.
Additional information:- Until a Maintenance code has been set in the "Define Maintenance code" parameter, allusers are automatically logged on in the Maintenance role. Once the Maintenance code
has been set, all users are automatically logged on in the Operator role.
- The access rights can be changed via the "Enter access code" parameter.

User interface• Operator• Maintenance• Service• Production• Development

Additional information

User interface• Operator option
Provides only read access to parameters.
• Maintenance option
Provides read and write access to parameters.
Additional information:
For some parameters, the user must be logged on in the Service role to obtain write
access.
• Service option
Provides read and write access to Service parameters.

Endress+Hauser

93

"System" menu

Proline Promag 10 Modbus RS485

Enter access code
Navigation

 System → User manag. → Ent. access code

Description

For users logged on in the Operator role, enter the Maintenance code to change the accessstatus to Maintenance and disable write protection of parameters. For users logged on in
the Maintenance role, enter the Service code to change the access status to Service andenable read and write access to Service parameters.

User entry

0 to 9 999

Reset Maintenance code
Navigation

 System → User manag. → Reset Maint code

Description

Enter the code provided by Endress+Hauser Technical Support to reset the Maintenancecode.

User entry

Character string comprising numbers, letters and special characters (32)94

Endress+Hauser

Proline Promag 10 Modbus RS485

5.2.1"System" menu

"Define access code" wizard

Complete this wizard to specify an access code for the Maintenance role.
Navigation

 System → User manag. → Def. access code

‣ Define Maintenance code
Define Maintenance code→  95

Confirm Maintenance code→  95

Define Maintenance code

Navigation

 System → User manag. → Def. Maint. code → Def. Maint. code

Description

Specify an access code that is required to obtain the access rights for the Maintenance role.

User entry

0 to 9 999

Confirm Maintenance code

Navigation

 System → User manag. → Def. Maint. code → Conf. Maint code

Description

Confirm the access code entered for the Maintenance role.

User entry

0 to 9 999

Endress+Hauser

95

"System" menu

Proline Promag 10 Modbus RS485

5.3"Connectivity" submenu

Navigation

 System → Connectivity‣ Connectivity‣ Bluetooth configuration

5.3.1→  96"Bluetooth configuration" submenu

Navigation

 System → Connectivity → Bluetooth conf.

‣ Bluetooth configuration
Bluetooth→  96

Communication established→  96

BluetoothNavigation

 System → Connectivity → Bluetooth conf. → Bluetooth

Description

Enable or disable Bluetooth.

Selection• Enable• Disable• Not available *Communication established
Navigation

 System → Connectivity → Bluetooth conf. → Communi. establ.

User interface

• No
• Yes

*

96

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485"System" menu

5.4

" Date / Time" submenu

Navigation

 System → Date / Time‣ Date/time
Set date/time→  97

Time format→  97

Time zone→  97

Set date/timeNavigation



System → Date/time → Set date/time

Description

Set the date and local time. Every time the date or time is changed, a logbook entry iscreated.

User entry

Date and time

Time format



Navigation

 System → Date/time → Time format

Description

Select time format.

Selection• 24 h• 12 h AM/PMTime zoneNavigation

 System → Date/time → Time zone

Description

Select the time zone. Every time the time zone is changed, a logbook entry is created.

Endress+Hauser

97

"System" menu

Selection

98

Proline Promag 10 Modbus RS485

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+14:00

Endress+Hauser

Proline Promag 10 Modbus RS485

5.5"System" menu

"Information" submenu

Navigation

 System → Information‣ Information

5.5.1
Navigation‣ Device→  99‣ Sensor electronic module (ISEM)→  102‣ Display module→  102"Device" submenu
 System → Information → Device‣ Device
Device name→  99Device tag→  100

Serial number→  100

Order code→  100

Firmware version→  100

Extended order code 1→  101

Extended order code 2→  101

Extended order code 3→  101

ENP version→  101

Manufacturer→  102

Device name
Navigation

 System → Information → Device → Device name

Description

Displays the name of the transmitter.
Additional information:
The name can also be found on the transmitter's nameplate.

Endress+Hauser

99

"System" menu

User interface

Proline Promag 10 Modbus RS485

Character string comprising numbers, letters and special characters

Device tag
Navigation

 System → Information → Device → Device tag

Description

Displays the name for the measuring point.

User interface

Character string comprising numbers, letters and special characters

Serial number
Navigation

 System → Information → Device → Serial number

Description

Displays the serial number of the measuring device. The serial number can be used to
identify the measuring device and to retrieve further information on the measuring device,
such as the related documentation, via the Device Viewer or Operations app.
Additional information:
The serial number can also be found on the nameplate of the sensor and transmitter.

User interface

Character string comprising numbers, letters and special characters

Order code

Navigation

 System → Information → Device → Order code

Description

Displays the device order code.
Additional information:
The order code can be used for instance to order a replacement or spare device or to verify
that the device features specified on the order form match the shipping note.

User interface

Character string comprising numbers, letters and special characters

Firmware version
Navigation

 System → Information → Device → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

100

Endress+Hauser

Proline Promag 10 Modbus RS485

Extended order code 1"System" menuNavigation

 System → Information → Device → Ext. order cd. 1

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Extended order code 2Navigation

 System → Information → Device → Ext. order cd. 2

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Extended order code 3

Navigation

 System → Information → Device → Ext. order cd. 3

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

ENP version
Navigation

 System → Information → Device → ENP version

Description

Displays the version of the electronic nameplate (ENP).

Endress+Hauser

101

"System" menu

User interface

Proline Promag 10 Modbus RS485

Character string comprising numbers, letters and special characters

Manufacturer
Navigation

 System → Information → Device → Manufacturer

Description

Displays the manufacturer.

User interface

Character string comprising numbers, letters and special characters

5.5.2"Sensor electronic module (ISEM)" submenu

Navigation

 System → Information → Sens. electronic‣ Sensor electronic module (ISEM)
Firmware version→  102

Firmware version
Navigation

 System → Information → Sens. electronic → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

5.5.3"Display module" submenu

Navigation

 System → Information → Display module‣ Display module
Firmware version102→  103

Endress+Hauser

Proline Promag 10 Modbus RS485"System" menu

Firmware version
Navigation

 System → Information → Display module → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

Endress+Hauser

103

"System" menu

Proline Promag 10 Modbus RS485

5.6"Display" submenu

Navigation

 System → Display‣ Display
Language→  104

Value 1 display→  105

Value 2 display→  105

Value 3 display→  106

Value 4 display→  106

Display damping→  106

Rotation display→  107

Brightness→  107

Color scheme→  107

Language
Navigation

 System → Display → Language

Description

Set display language.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• ‫( الْعَرَبيّة‬Arabic) *• Bahasa Indonesia*

104

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485"System" menu• ภาษาไทย (Thai) *• tiếng Việt (Vietnamese)• čeština (Czech)

Value 1 display

Navigation

 System → Display → Value 1 display

Description

Select the measured value that is displayed first on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 2 display

Navigation

 System → Display → Value 2 display

Description

Select the measured value that is shown second on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection*• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Visibility depends on order options or device settings

Endress+Hauser

105

"System" menu

Proline Promag 10 Modbus RS485

Value 3 display

Navigation

 System → Display → Value 3 display

Description

Select the measured value that is shown third on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 4 display

Navigation

 System → Display → Value 4 display

Description

Select the measured value that is shown fourth on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Display damping

Navigation

 System → Display → Display damping

Description

Enter time constant (PT1 element) to set reaction time of the display to fluctuations in themeasured value.
Additional information:- The smaller the time constant the faster the display reacts to fluctuations in themeasured value.
- If the time constant is set to 0, damping is deactivated.

*

106

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485

User entry"System" menu

0.0 to 999.9 s

Rotation display

Navigation

 System → Display → Rotation display

Description

Select rotation angle of the display text to optimize local display readability.

Selection• Auto• 0 degree• 90 degree• 180 degree• 270 degree

Brightness
Navigation

 System → Display → Brightness

Description

Adjust brightness.

User entry

0 to 100 %Color scheme

Navigation

 System → Display → Color scheme

Description

Select preferred color scheme.

Selection• Light• DarkEndress+Hauser

107

"System" menu

Proline Promag 10 Modbus RS485

5.7"Software configuration" submenu

Navigation

 System → Software config.

‣ Software configuration
Activate SW option→  108

Software option overview→  108

Activate SW option



Navigation

 System → Software config. → Activate SW opt.

Description

Enter application package code or code of the functionality ordered separately to activate
it.
Additional information:- If a measuring device was ordered with an add-on software option, the activation code isprogrammed into the measuring device ex factory.
- After entering the activation code: Check whether the new software option is displayedin the "Software option overview" parameter and therefore active.
NOTE
If an an invalid code is entered the software options that have already been activated areinvalidated!
Before entering a new activation code: Create a record of the existing activation code.

User entry

Positive integer

Software option overview
Navigation

 System → Software config. → SW option overv.

Description

Displays all software options included in the order ex factory or ordered at a later date thathave been enabled via the operating interface.
Additional information:
If a new software option is not displayed after entering the activation code, the code
entered was inaccurate or invalid. In this case, contact the appropriate Endress+Hauser
sales organization to activate the software option.

User interface

108• Heartbeat Verification• Heartbeat Monitoring

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

6

Modbus RS485 register information

6.1

Notes

6.1.1

Structure of the register information

The individual parts of a parameter description are described in the following section:
Navigation: navigation path to the parameter
Parameter

Register

Data type

Access type

Selection/input→Name of parameter

Indicated in decimal
numerical format• Float length =4 byte• Integer length =2 byte• String length,
depending onparameter

Possible type of accessto parameter:• Read access via
function codes 03,
04 or 23
• Write access via
function codes 06,
16 or 23

Selection
List of the individual
options for theparameter• Option 1• Option 2• Option 3 (+)

Page number informationand cross-reference to thestandard parameter
description • Factory
settinghighlighted inbold• (+) = Factory
setting
depends on
country, order
options ordevicesettings
Input
Input range for theparameter

NOTICE
If non-volatile device parameters are modified via the MODBUS RS485 function codes
06, 16 or 23, the change is saved in the EEPROM of the measuring device.
The number of writes to the EEPROM is technically restricted to a maximum of 1 million.
‣ Make sure to comply with this limit since, if it is exceeded, data loss and measuring
device failure will result.
‣ Avoid constantly writing non-volatile device parameters via the MODBUS RS485.

6.1.2

Address model

The Modbus RS485 register addresses of the measuring device are implemented inaccordance with the "Modbus Applications Protocol Specification V1.1".
In addition, systems are used that work with the register address model "Modicon Modbus
Protocol Reference Guide (PI-MBUS-300 Rev. J)".
Depending on the function code used, a number is added at the start of the register
address with this specification:• "3" → "Read" access• "4" → "Write" accessEndress+Hauser

109Modbus RS485 register information

Proline Promag 10 Modbus RS485

Function code Access type Register in accordance with "Modbus
Applications Protocol Specification"

Register in accordance with
"Modicon Modbus Protocol
Reference Guide"

03
04
23

Read

XXXX
Example: mass flow = 2007

3XXXX
Example: mass flow = 32007

06
16
23

Write

XXXX
Example: reset totalizer = 6401

4XXXX
Example: reset totalizer = 46401

6.2

Overview of the operating menu

The following table provides an overview of the menu structure of the operating menu andits parameters. The page reference indicates where the associated description of thesubmenu or parameter can be found.
→  119

Guidance‣ Commissioning110→  119
Device tag→  119

Serial number→  119

Firmware version→  119

Device name→  119

Volume flow unit

→  120

Volume unit

→  121Mass flow unit

→  122

Mass unit→  122

Density unit

→  122

Temperature unit

→  122

Conductivity unit

→  122

Assign process variable→  123

Unit totalizer 1 to n

→  123

Totalizer operation mode→  123

Failure mode→  123

Flow damping→  123

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Low flow cut off→  123

On value low flow cutoff→  123

Off value low flow cutoff→  123

Pressure shock suppression→  123

Empty pipe detection→  123

Empty pipe adjust value→  123

Full pipe adjust value→  123

Process variable current output→  123

Current range output→  124

Lower range value output→  124

Upper range value output→  124

Damping current output→  124

Fixed current→  124

Failure behavior current output→  124

Failure current→  124

Value 1 display→  124

Value 2 display→  124

Value 3 display→  124

Value 4 display→  124

Display damping→  124

Time format→  125

Time zone→  125→  125

Diagnostics‣ Active diagnostics
Actual diagnostics

Endress+Hauser→  125→  125111

Modbus RS485 register information

Proline Promag 10 Modbus RS485

Timestamp→  125

Previous diagnostics→  125

Timestamp→  125

Operating time from restart→  126

Operating time→  126‣ Diagnostic list

→  126‣ Simulation→  126
Assign simulation process variable→  126

Process variable value→  126

Current output 1 simulation→  126

Current output value→  126

Device alarm simulation→  126

Diagnostic event simulation→  126‣ Heartbeat Technology→  126‣ Diagnostic settings→  126‣ Properties→  126→  126

Alarm delay‣ Diagnostic configuration→  126‣ Sensor→  126
Assign behavior of diagnostic no. 043‣ Electronics112→  126→  127

Assign behavior of diagnostic no. 230→  127

Assign behavior of diagnostic no. 231→  127

Assign behavior of diagnostic no. 302→  127

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Assign behavior of diagnostic no. 376→  127

Assign behavior of diagnostic no. 377→  127‣ Configuration→  127
Assign behavior of diagnostic no. 441‣ Process→  127
Assign behavior of diagnostic no. 832→  127

Assign behavior of diagnostic no. 833→  127

Assign behavior of diagnostic no. 834→  127

Assign behavior of diagnostic no. 835→  127

Assign behavior of diagnostic no. 842→  127

Assign behavior of diagnostic no. 937→  128

Assign behavior of diagnostic no. 938→  128

Assign behavior of diagnostic no. 961→  128

Assign behavior of diagnostic no. 962→  128→  128

Application‣ Measured values

Endress+Hauser→  127→  128
Mass flow→  128

Volume flow

→  128

Conductivity→  128

Corrected conductivity→  128

Temperature→  128‣ Totalizer→  128
Totalizer value 1 to n

→  128

Totalizer overflow 1 to n

→  128113

Modbus RS485 register information

Proline Promag 10 Modbus RS485‣ System units→  129
Volume flow unit

→  129

Volume unit

→  130

Mass flow unit

→  131

Mass unit→  131

Density unit

→  131

Temperature unit

→  131

Conductivity unit

→  131‣ Totalizers→  132‣ Totalizer handlingReset all totalizers‣ Totalizer 1 to n

→  132→  132

Assign process variable→  132

Unit totalizer 1 to n

→  132

Totalizer operation mode→  132

Control Totalizer 1 to n

→  132

Preset value 1 to n

→  132

Failure mode→  132‣ Sensor→  133‣ Process parameters114→  132→  133

Flow damping→  133

Flow override→  133

Conductivity measurement→  133

Conductivity temperature coefficient→  133

Fixed density→  133

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information‣ Low flow cut off→  133
Low flow cut off→  133

On value low flow cutoff→  133

Off value low flow cutoff→  133‣ Empty pipe detection
Empty pipe detection→  133

New adjustment→  133

Empty pipe adjust value→  133

Full pipe adjust value→  133

Measured value EPD→  133‣ Sensor adjustment→  133

Installation direction→  133

Integration time→  133

Measuring period→  133‣ Calibration→  134
Nominal diameter→  134

Calibration factor→  134

Zero point→  134

Conductivity calibration factor→  134‣ Electrode cleaning cycle

Endress+Hauser→  133→  134

Electrode cleaning cycle→  134

ECC duration→  134

ECC recovery time→  134

ECC interval→  134

ECC polarity→  134

115

Modbus RS485 register information

Proline Promag 10 Modbus RS485‣ Current output 1→  134
Process variable current output→  134

Measuring mode current output→  134

Current range output→  134

Fixed current→  134

Lower range value output→  134

Upper range value output→  134

Damping current output→  134

Failure behavior current output→  134

Failure current→  135

Output current 1→  135‣ Communication→  135‣ Modbus configuration116→  135

Bus address→  135

Baudrate→  135

Parity→  135

Byte order→  135

Telegram delay→  135

Failure mode→  135

Fieldbus writing access→  135

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information‣ Modbus data map
Scan list register 0 to 15‣ Modbus information→  135→  136

Device ID→  136

Device revision→  136→  136

System‣ Device management→  136

Device tag→  136

Locking status→  136

Configuration counter→  136

Device reset→  136‣ User management→  136

User role→  136

Enter access code→  136

Reset Maintenance code→  136‣ Define Maintenance code→  136

Define Maintenance code→  136

Confirm Maintenance code→  136‣ Connectivity→  137‣ Bluetooth configuration‣ Date/time

Endress+Hauser→  135→  137

Bluetooth→  137

Communication established→  137→  137

Time format→  137

Time zone→  137117

Modbus RS485 register information

Proline Promag 10 Modbus RS485‣ Information→  138‣ Device→  138
Device name→  138

Device tag→  138

Serial number→  138

Order code→  138

Firmware version→  138

Extended order code 1→  138

Extended order code 2→  138

Extended order code 3→  138

ENP version→  138

Manufacturer→  138‣ Sensor electronic module (ISEM)→  138

Firmware version→  138‣ Display module→  138
Firmware version‣ Display118→  138→  139

Language→  139

Value 1 display→  139

Value 2 display→  139

Value 3 display→  139

Value 4 display→  139

Display damping→  139

Rotation display→  139

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Brightness→  140

Color scheme→  140‣ Software configuration→  140

Activate SW option→  140

Software option overview→  140

6.3

Register information

6.3.1"Guidance" menu

"Commissioning" wizard
Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Device tag

29243 to 29258

StringRead / Write

Character string comprising numbers,letters and special characters (32)8

Serial number

7003 to 7008

StringRead

Character string comprising numbers,letters and special characters8

Firmware version

7277 to 7280

StringRead

Character string comprising numbers,letters and special characters

8Device name7263 to 7270

StringRead

Character string comprising numbers,letters and special characters

8Endress+Hauser

119Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface

Volume flow unit

2103

Integer

Read / Write

0 = cm³/s
1 = cm³/min
2 = cm³/h
3 = cm³/d
4 = dm³/s
5 = dm³/min
6 = dm³/h
7 = dm³/d
8 = m³/s
9 = m³/min
10 = m³/h
11 = m³/d
12 = ml/s
13 = ml/min
14 = ml/h
15 = ml/d
16 = l/s
17 = l/min
18 = l/h
19 = l/d
20 = hl/s
21 = hl/min
22 = hl/h
23 = hl/d
24 = Ml/s
25 = Ml/min
26 = Ml/h
27 = Ml/d
32 = af/s
33 = af/min
34 = af/h
35 = af/d
36 = ft³/s
37 = ft³/min
38 = ft³/h
39 = ft³/d
40 = fl oz/s (us)41 = fl oz/min (us)42 = fl oz/h (us)43 = fl oz/d (us)44 = gal/s (us)45 = gal/min (us)46 = gal/h (us)47 = gal/d (us)48 = Mgal/s (us)49 = Mgal/min (us)50 = Mgal/h (us)51 = Mgal/d (us)52 = bbl/s (us;liq.)
53 = bbl/min (us;liq.)
54 = bbl/h (us;liq.)
55 = bbl/d (us;liq.)
56 = bbl/s (us;beer)
57 = bbl/min (us;beer)58 = bbl/h (us;beer)59 = bbl/d (us;beer)
60 = bbl/s (us;oil)
61 = bbl/min (us;oil)
62 = bbl/h (us;oil)
63 = bbl/d (us;oil)
64 = bbl/s (us;tank)65 = bbl/min (us;tank)66 = bbl/h (us;tank)
67 = bbl/d (us;tank)
68 = gal/s (imp)
69 = gal/min (imp)
70 = gal/h (imp)

120

→ 
9

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

71 = gal/d (imp)
72 = Mgal/s (imp)
73 = Mgal/min (imp)
74 = Mgal/h (imp)
75 = Mgal/d (imp)
76 = bbl/s (imp;beer)
77 = bbl/min (imp;beer)
78 = bbl/h (imp;beer)
79 = bbl/d (imp;beer)
80 = bbl/s (imp;oil)
81 = bbl/min (imp;oil)
82 = bbl/h (imp;oil)
83 = bbl/d (imp;oil)
88 = kgal/s (us)89 = kgal/min (us)90 = kgal/h (us)91 = kgal/d (us)92 = MMft³/s
93 = MMft³/min
94 = MMft³/h
96 = Mft³/d
Volume unit

Endress+Hauser

2104

Integer

Read / Write

0 = cm³
1 = dm³
2 = m³3 = ml4=l
5 = hl
6 = Ml Mega
8 = af
9 = ft³
10 = fl oz (us)11 = gal (us)12 = Mgal (us)13 = bbl (us;liq.)
14 = bbl (us;beer)
15 = bbl (us;oil)
16 = bbl (us;tank)
17 = gal (imp)
18 = Mgal (imp)
19 = bbl (imp;beer)
20 = bbl (imp;oil)
22 = kgal (us)23 = Mft³

10121

Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Mass flow unit

2101

Integer

Read / Write

0 = g/s
1 = g/min
2 = g/h
3 = g/d
4 = kg/s
5 = kg/min
6 = kg/h
7 = kg/d
8 = t/s
9 = t/min
10 = t/h
11 = t/d
12 = oz/s
13 = oz/min
14 = oz/h
15 = oz/d
16 = lb/s
17 = lb/min
18 = lb/h
19 = lb/d
20 = STon/s
21 = STon/min
22 = STon/h
23 = STon/d11

Mass unit

2102

Integer

Read / Write

50 = g51 = kg
52 = t53 = oz
54 = lb
55 = STon

11

Density unit

2107

Integer

Read / Write

0 = g/cm³
2 = kg/dm³
3 = kg/l
4 = kg/m³
5 = SD4°C
6 = SD15°C
7 = SD20°C
8 = SG4°C
9 = SG15°C
10 = SG20°C
11 = lb/ft³
12 = lb/gal (us)13 = lb/bbl (us;liq.)
14 = lb/bbl (us;beer)
15 = lb/bbl (us;oil)
16 = lb/bbl (us;tank)
17 = lb/gal (imp)
18 = lb/bbl (imp;beer)
19 = lb/bbl (imp;oil)
21 = g/m³

12Temperature unit

2109

Integer

Read / Write

0 = °C
1=K
2 = °F
3 = °R

12

Conductivity unit

2121

Integer

Read / Write

1 = MS/m
2 = kS/m
3 = S/m
4 = S/cm
5 = mS/m
6 = mS/cm
7 = µS/m
8 = µS/cm
9 = µS/mm
10 = nS/cm

12

122

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign process variable

2601

Integer

Read / Write

0 = Off
1 = Volume flow
11 = Mass flow

13

Unit totalizer 1 to n

1: 4604
2: 4605
3: 4606

Integer

Read / Write

0 = cm³ *1 = dm³ *2 = m³ *3 = ml *4 = l*
5 = hl *6 = Ml Mega *8 = af *9 = ft³ *10 = fl oz (us) *11 = gal (us) *12 = Mgal (us) *13 = bbl (us;liq.) *14 = bbl (us;beer) *15 = bbl (us;oil) *16 = bbl (us;tank) *17 = gal (imp) *18 = Mgal (imp) *19 = bbl (imp;beer) *20 = bbl (imp;oil) *22 = kgal (us) *23 = Mft³ *
50 = g *51 = kg *52 = t *53 = oz *54 = lb *55 = STon *251 = None *13

Totalizer operation mode

2605

Integer

Read / Write

0 = Net flow total
1 = Forward flow total
2 = Reverse flow total

14

Failure mode

2606

Integer

Read / Write

0 = Stop
1 = Actual value
2 = Last valid value

15

Flow damping

2274

Integer

Read / Write

0 to 15

15

Low flow cut off

5101

Integer

Read / Write

0 = Off
1 = Volume flow
11 = Mass flow

17

On value low flow cutoff

5138 to 5139

Float

Read / Write

Positive floating-point number18

Off value low flow cutoff

5104 to 5105

Float

Read / Write

0 to 100.0 %18

Pressure shock suppression

5140 to 5141

Float

Read / Write

0 to 100 s

16

Empty pipe detection

5106

Integer

Read / Write

0 = Off
1 = On19

Empty pipe adjust value

2181 to 2182

Float

Read

Positive floating-point number

19

Full pipe adjust value

2832 to 2833

Float

Read

Positive floating-point number

19

Process variable current output

5927

Integer

Read / Write

0 = Off *1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
40 = Noise *41 = Coil current shot time *20

Endress+Hauser

123Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Current range output

5923

Integer

Read / Write

0 = 4...20 mA (4... 20.5 mA)
1 = 4...20 mA US (3.9...20.8 mA)
2 = 4...20 mA NE (3.8...20.5 mA)
4 = Fixed value

20

Lower range value output

6195 to 6196

Float

Read / Write

Signed floating-point number

21

Upper range value output

5915 to 5916

Float

Read / Write

Signed floating-point number

22

Damping current output

5903 to 5904

Float

Read / Write

0.0 to 999.9 s

23

Fixed current

5987 to 5988

Float

Read / Write

3.59 to 21.5 mA

23

Failure behavior current output

5911

Integer

Read / Write

0 = Min.
1 = Max.
4 = Actual value
5 = Last valid value
6 = Fixed value

23

Failure current

5979 to 5980

Float

Read / Write

3.59 to 21.5 mA

24

Value 1 display

34918

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3

25

Value 2 display

34919

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3
251 = None

25

Value 3 display

34922

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3
251 = None

26

Value 4 display

34923

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3
251 = None26

Display damping

27602 to 27603

Float

Read / Write

0.0 to 999.9 s

26

124

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Navigation: Guidance → Commissioning
Parameter

Register

Data type

AccessSelection / User entry / User interface→ Time format

2150

Integer

Read / Write

12 = 12 h AM/PM
24 = 24 h

27Time zone

27339

Integer

Read / Write

0 = UTC 00:00
1 = UTC+01:00
2 = UTC+02:00
3 = UTC+03:00
4 = UTC+04:00
5 = UTC+05:00
6 = UTC+06:00
7 = UTC+07:00
8 = UTC+08:00
9 = UTC+09:00
10 = UTC+10:00
11 = UTC+11:00
12 = UTC+12:00
13 = UTC+13:00
14 = UTC+14:00
35 = UTC+03:30
45 = UTC+04:30
55 = UTC+05:30
57 = UTC+05:45
65 = UTC+06:30
87 = UTC+08:45
95 = UTC+09:30
105 = UTC+10:30
127 = UTC+12:45
135 = UTC-03:30
195 = UTC-09:30
201 = UTC-01:00
202 = UTC-02:00
203 = UTC-03:00
204 = UTC-04:00
205 = UTC-05:00
206 = UTC-06:00
207 = UTC-07:00
208 = UTC-08:00
209 = UTC-09:00
210 = UTC-10:00
211 = UTC-11:00
212 = UTC-12:00

27

*
*
*

Visibility depends on order options or device settings
Visibility depends on order options or device settings
Visibility depends on order options or device settings

6.3.2"Diagnostics" menu

"Active diagnostics" submenu
Navigation: Diagnostics → Active diagnostics
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Actual diagnostics

2732

Integer

Read

Positive integer

30

Timestamp

29726

StringReadDays (d), hours (h), minutes (m), seconds(s)30Previous diagnostics

2734

Integer

Read

Positive integer

30

Timestamp

29715

StringReadDays (d), hours (h), minutes (m), seconds(s)31

Endress+Hauser

125Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: Diagnostics → Active diagnostics
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Operating time from restart

2624

StringReadDays (d), hours (h), minutes (m), seconds(s)31

Operating time

2631

StringReadDays (d), hours (h), minutes (m), seconds(s)31"Diagnostic list" submenu"Simulation" submenu
Navigation: Diagnostics → Simulation
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign simulation process variable

6813

Integer

Read / Write

0 = Off
1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow

35

Process variable value

6814 to 6815

Float

Read / Write

Signed floating-point number

36

Current output 1 simulation

5939

Integer

Read / Write

0 = Off
1 = On36

Current output value

5995 to 5996

Float

Read / Write

3.59 to 21.5 mA

36

Device alarm simulation

6812

Integer

Read / Write

0 = Off
1 = On37

Diagnostic event simulation

4259

Integer

Read / Write

0 = Off

37*

Visibility depends on order options or device settings"Heartbeat Technology" submenu"Diagnostic settings" submenu"Properties" submenu
Navigation: Diagnostics → Diagnostic settings → Properties
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Alarm delay

6808 to 6809

Float

Read / Write

0 to 60 s

39"Diagnostic configuration" submenu"Sensor" submenu
Navigation: Diagnostics → Diagnostic settings → Diagnostic configuration → Sensor
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign behavior of diagnostic no.
043

9666

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

40

126

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information"Electronics" submenu
Navigation: Diagnostics → Diagnostic settings → Diagnostic configuration → Electronics
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign behavior of diagnostic no.
230

27632

Integer

Read / Write

1 = Logbook entry only
2 = Warning
3 = Alarm

41

Assign behavior of diagnostic no.
231

28535

Integer

Read / Write

1 = Logbook entry only
2 = Warning
3 = Alarm

41

Assign behavior of diagnostic no.
302

6484

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning

41

Assign behavior of diagnostic no.
376

6442

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

41

Assign behavior of diagnostic no.
377

5183

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

42"Configuration" submenu
Navigation: Diagnostics → Diagnostic settings → Diagnostic configuration → Configuration
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign behavior of diagnostic no.
441

4742

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

43"Process" submenu
Navigation: Diagnostics → Diagnostic settings → Diagnostic configuration → Process
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign behavior of diagnostic no.
832

6440

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

44

Assign behavior of diagnostic no.
833

6439

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

44

Assign behavior of diagnostic no.
834

6438

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

45

Assign behavior of diagnostic no.
835

6437

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

45

Assign behavior of diagnostic no.
842

9661

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm46

Endress+Hauser

127Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: Diagnostics → Diagnostic settings → Diagnostic configuration → Process
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign behavior of diagnostic no.
937

26404

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm46

Assign behavior of diagnostic no.
938

5837

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

47

Assign behavior of diagnostic no.
961

28459

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

47

Assign behavior of diagnostic no.
962

6441

Integer

Read / Write

0 = Off
1 = Logbook entry only
2 = Warning
3 = Alarm

48

6.3.3

"Application" menu

"Measured values" submenu
Navigation: Application → Measured values
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Mass flow

2007 to 2008

Float

Read

Signed floating-point number

49

Volume flow

2009 to 2010

Float

Read

Signed floating-point number

50

Conductivity

2099 to 2100

Float

Read

Positive floating-point number

50

Corrected conductivity

37423 to 37424

Float

Read

Positive floating-point number

50

Temperature2017 to 2018

Float

Read

Positive floating-point number

51"Totalizer" submenu
Navigation: Application → Measured values → Totalizer
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Totalizer value 1 to n

1: 2610 to 2611
2: 2810 to 2811
3: 3010 to 3011

Float

Read

Signed floating-point number

51

Totalizer overflow 1 to n

1: 2612 to 2613
2: 2812 to 2813
3: 3012 to 3013

Float

Read

–32 000.0 to 32 000.0

52

128

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information"System units" submenu
Navigation: Application → System units
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Volume flow unit

2103

Integer

Read / Write

0 = cm³/s
1 = cm³/min
2 = cm³/h
3 = cm³/d
4 = dm³/s
5 = dm³/min
6 = dm³/h
7 = dm³/d
8 = m³/s
9 = m³/min
10 = m³/h
11 = m³/d
12 = ml/s
13 = ml/min
14 = ml/h
15 = ml/d
16 = l/s
17 = l/min
18 = l/h
19 = l/d
20 = hl/s
21 = hl/min
22 = hl/h
23 = hl/d
24 = Ml/s
25 = Ml/min
26 = Ml/h
27 = Ml/d
32 = af/s
33 = af/min
34 = af/h
35 = af/d
36 = ft³/s
37 = ft³/min
38 = ft³/h
39 = ft³/d
40 = fl oz/s (us)41 = fl oz/min (us)42 = fl oz/h (us)43 = fl oz/d (us)44 = gal/s (us)45 = gal/min (us)46 = gal/h (us)47 = gal/d (us)48 = Mgal/s (us)49 = Mgal/min (us)50 = Mgal/h (us)51 = Mgal/d (us)52 = bbl/s (us;liq.)
53 = bbl/min (us;liq.)
54 = bbl/h (us;liq.)
55 = bbl/d (us;liq.)
56 = bbl/s (us;beer)
57 = bbl/min (us;beer)58 = bbl/h (us;beer)59 = bbl/d (us;beer)
60 = bbl/s (us;oil)
61 = bbl/min (us;oil)
62 = bbl/h (us;oil)
63 = bbl/d (us;oil)
64 = bbl/s (us;tank)65 = bbl/min (us;tank)66 = bbl/h (us;tank)
67 = bbl/d (us;tank)
68 = gal/s (imp)53

Endress+Hauser

129Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: Application → System units
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

69 = gal/min (imp)
70 = gal/h (imp)
71 = gal/d (imp)
72 = Mgal/s (imp)
73 = Mgal/min (imp)
74 = Mgal/h (imp)
75 = Mgal/d (imp)
76 = bbl/s (imp;beer)
77 = bbl/min (imp;beer)
78 = bbl/h (imp;beer)
79 = bbl/d (imp;beer)
80 = bbl/s (imp;oil)
81 = bbl/min (imp;oil)
82 = bbl/h (imp;oil)
83 = bbl/d (imp;oil)
88 = kgal/s (us)89 = kgal/min (us)90 = kgal/h (us)91 = kgal/d (us)92 = MMft³/s
93 = MMft³/min
94 = MMft³/h
96 = Mft³/d
Volume unit

130

2104

Integer

Read / Write

0 = cm³
1 = dm³
2 = m³3 = ml4=l
5 = hl
6 = Ml Mega
8 = af
9 = ft³
10 = fl oz (us)11 = gal (us)12 = Mgal (us)13 = bbl (us;liq.)
14 = bbl (us;beer)
15 = bbl (us;oil)
16 = bbl (us;tank)
17 = gal (imp)
18 = Mgal (imp)
19 = bbl (imp;beer)
20 = bbl (imp;oil)
22 = kgal (us)23 = Mft³

54

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Navigation: Application → System units
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Mass flow unit

2101

Integer

Read / Write

0 = g/s
1 = g/min
2 = g/h
3 = g/d
4 = kg/s
5 = kg/min
6 = kg/h
7 = kg/d
8 = t/s
9 = t/min
10 = t/h
11 = t/d
12 = oz/s
13 = oz/min
14 = oz/h
15 = oz/d
16 = lb/s
17 = lb/min
18 = lb/h
19 = lb/d
20 = STon/s
21 = STon/min
22 = STon/h
23 = STon/d55

Mass unit

2102

Integer

Read / Write

50 = g51 = kg
52 = t53 = oz
54 = lb
55 = STon

55

Density unit

2107

Integer

Read / Write

0 = g/cm³
2 = kg/dm³
3 = kg/l
4 = kg/m³
5 = SD4°C
6 = SD15°C
7 = SD20°C
8 = SG4°C
9 = SG15°C
10 = SG20°C
11 = lb/ft³
12 = lb/gal (us)13 = lb/bbl (us;liq.)
14 = lb/bbl (us;beer)
15 = lb/bbl (us;oil)
16 = lb/bbl (us;tank)
17 = lb/gal (imp)
18 = lb/bbl (imp;beer)
19 = lb/bbl (imp;oil)
21 = g/m³

56

Temperature unit

2109

Integer

Read / Write

0 = °C
1=K
2 = °F
3 = °R

56

Conductivity unit

2121

Integer

Read / Write

1 = MS/m
2 = kS/m
3 = S/m
4 = S/cm
5 = mS/m
6 = mS/cm
7 = µS/m
8 = µS/cm
9 = µS/mm
10 = nS/cm

56

Endress+Hauser

131Modbus RS485 register information

Proline Promag 10 Modbus RS485"Totalizers" submenu"Totalizer handling" submenu
Navigation: Application → Totalizers → Totalizer handling
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Reset all totalizers

2609

Integer

Read / Write

0 = Cancel
1 = Reset + totalize

58"Totalizer 1 to n" submenu
Navigation: Application → Totalizers → Totalizer 1 to n
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Assign process variable

1: 2601
2: 2801
3: 3001

Integer

Read / Write

0 = Off
1 = Volume flow
11 = Mass flow

59

Unit totalizer 1 to n

1: 4604
2: 4605
3: 4606

Integer

Read / Write

0 = cm³ *1 = dm³ *2 = m³ *3 = ml *4 = l*
5 = hl *6 = Ml Mega *8 = af *9 = ft³ *10 = fl oz (us) *11 = gal (us) *12 = Mgal (us) *13 = bbl (us;liq.) *14 = bbl (us;beer) *15 = bbl (us;oil) *16 = bbl (us;tank) *17 = gal (imp) *18 = Mgal (imp) *19 = bbl (imp;beer) *20 = bbl (imp;oil) *22 = kgal (us) *23 = Mft³ *
50 = g *51 = kg *52 = t *53 = oz *54 = lb *55 = STon *251 = None *59

Totalizer operation mode

1: 2605
2: 2805
3: 3005

Integer

Read / Write

0 = Net flow total
1 = Forward flow total
2 = Reverse flow total

60

Control Totalizer 1 to n

1: 2608
2: 2808
3: 3008

Integer

Read / Write

0 = Totalize
1 = Reset + totalize
2 = Preset + hold
3 = Reset + hold
5 = Hold60

Preset value 1 to n

1: 2590 to 2591
2: 2592 to 2593
3: 2594 to 2595

Float

Read / Write

Signed floating-point number

61

Failure mode

1: 2606
2: 2806
3: 3006

Integer

Read / Write

0 = Stop
1 = Actual value
2 = Last valid value

61

*

132

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information"Sensor" submenu"Process parameters" submenu
Navigation: Application → Sensor → Process parameters
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Flow damping

2274

Integer

Read / Write

0 to 1564

Flow override

5503

Integer

Read / Write

0 = Off
1 = On64

Conductivity measurement

2268

Integer

Read / Write

0 = Off
1 = On64

Conductivity temperature coefficient 2886 to 2887

Float

Read / Write

Signed floating-point number

65

Fixed density

Float

Read / Write

Positive floating-point number

66

2830 to 2831

"Low flow cut off" submenu
Navigation: Application → Sensor → Low flow cut off
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Low flow cut off

5101

Integer

Read / Write

0 = Off
1 = Volume flow
11 = Mass flow

66

On value low flow cutoff

5138 to 5139

Float

Read / Write

Positive floating-point number

67

Off value low flow cutoff

5104 to 5105

Float

Read / Write

0 to 100.0 %67"Empty pipe detection" submenu
Navigation: Application → Sensor → Empty pipe detection
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Empty pipe detection

5106

Integer

Read / Write

0 = Off
1 = On68

New adjustment

2335

Integer

Read / Write

0 = Cancel
1 = Empty pipe adjust
2 = Full pipe adjust

68

Empty pipe adjust value

2181 to 2182

Float

Read

Positive floating-point number

68

Full pipe adjust value

2832 to 2833

Float

Read

Positive floating-point number

69

Measured value EPD

2298 to 2299

Float

Read

Positive floating-point number

69"Sensor adjustment" submenu
Navigation: Application → Sensor → Sensor adjustment
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Installation direction

5501

Integer

Read / Write

0 = Forward flow
1 = Reverse flow

69

Integration time

2260 to 2261

Float

Read

1 to 65 ms

70

Measuring period

2852 to 2853

Float

Read

0 to 1 000 ms

70

Endress+Hauser

133

Modbus RS485 register information

Proline Promag 10 Modbus RS485"Calibration" submenu
Navigation: Application → Sensor → Calibration
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Nominal diameter

2048 to 2057

StringRead

Character string comprising numbers,letters and special characters

70

Calibration factor

2313 to 2314

Float

Read

Positive floating-point number

71

Zero point

2870 to 2871

Float

Read

Signed floating-point number

71

Conductivity calibration factor

19806 to 19807

Float

Read

0.01 to 10 000

71"Electrode cleaning cycle" submenu
Navigation: Application → Sensor → Electrode cleaning cycle
Parameter

Register

Data type

AccessSelection / User entry / User interface→ Electrode cleaning cycle

2280

Integer

Read / Write

0 = Off
1 = On72

ECC duration

2330 to 2331

Float

Read / Write

0.01 to 30 s

72

ECC recovery time

2332 to 2333

Float

Read / Write

1 to 600 s

72

ECC interval

2328 to 2329

Float

Read / Write

0.5 to 168 h

73

ECC polarity

2334

Integer

Read

0 = Positive
1 = Negative

73

AccessSelection / User entry / User interface→ "Current output 1" submenu
Navigation: Application → Current output 1
Parameter

Register

Data type*

Process variable current output

5927

Integer

Read / Write

0 = Off
1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
40 = Noise *41 = Coil current shot time *74

Measuring mode current output

5899

Integer

Read / Write

0 = Forward flow
2 = Reverse flow compensation
13 = Forward/Reverse flow *75

Current range output

5923

Integer

Read / Write

0 = 4...20 mA (4... 20.5 mA)
1 = 4...20 mA US (3.9...20.8 mA)
2 = 4...20 mA NE (3.8...20.5 mA)
4 = Fixed value

79

Fixed current

5987 to 5988

Float

Read / Write

3.59 to 21.5 mA

80

Lower range value output

6195 to 6196

Float

Read / Write

Signed floating-point number

80

Upper range value output

5915 to 5916

Float

Read / Write

Signed floating-point number

82

Damping current output

5903 to 5904

Float

Read / Write

0.0 to 999.9 s

82

Failure behavior current output

5911

Integer

Read / Write

0 = Min.
1 = Max.
4 = Actual value
5 = Last valid value
6 = Fixed value

83

134

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information

Navigation: Application → Current output 1
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Failure current

5979 to 5980

Float

Read / Write

3.59 to 21.5 mA

83

Output current 1

5931 to 5932

Float

Read

3.59 to 21.5 mA

84*

Visibility depends on order options or device settings"Communication" submenu"Modbus configuration" submenu
Navigation: Application → Communication → Modbus configuration
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Bus address

4910

Integer

Read / Write

1 to 247

85

Baudrate

4912

Integer

Read / Write

0 = 1200 BAUD
1 = 2400 BAUD
2 = 4800 BAUD
3 = 9600 BAUD
4 = 19200 BAUD
5 = 38400 BAUD
6 = 57600 BAUD
7 = 115200 BAUD

86

Parity4914

Integer

Read / Write

0 = Even
1 = Odd
2 = None / 2 stop bits
3 = None / 1 stop bit

86

Byte order4915

Integer

Read / Write

0 = 0-1-2-3
1 = 3-2-1-0
2 = 2-3-0-1
3 = 1-0-3-2

87

Telegram delay

4916 to 4917

Float

Read / Write

0 to 100 ms

87

Failure mode

4920

Integer

Read / Write

1 = Last valid value
255 = NaN value

87

Fieldbus writing access

6807

Integer

Read / Write

0 = Read + write
1 = Read only88

"Modbus data map" submenu
Navigation: Application → Communication → Modbus data map
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Scan list register 0 to 15

0: 5001
1: 5002
2: 5003
3: 5004
4: 5005
5: 5006
6: 5007
7: 5008
8: 5009
9: 5010
10: 5011
11: 5012
12: 5013
13: 5014
14: 5015
15: 5016

Integer

Read / Write

0 to 65 535

88

Endress+Hauser

135Modbus RS485 register information

Proline Promag 10 Modbus RS485"Modbus information" submenu
Navigation: Application → Communication → Modbus information
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Device ID

2547

Integer

Read

0 to 65 535

89

Device revision

4481

Integer

Read

0 to 65 535

89

6.3.4

"System" menu

"Device management" submenu
Navigation: System → Device management
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Device tag

29243 to 29258

StringRead / Write

Character string comprising numbers,letters and special characters (32)91

Locking status

4918

Integer

Read

256 = Hardware locked
512 = Temporarily locked

91

Configuration counter

4818

Integer

Read

0 to 65 535

92

Device reset

6817

Integer

Read / Write

0 = Cancel
1 = Restart device
2 = To delivery settings
25 = Restore S-DAT backup *35 = Restore T-DAT backup *36 = Create T-DAT backup

92*

Visibility depends on order options or device settings"User management" submenu
Navigation: System → User management
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

User role

2178

Integer

Read

0 = Operator
1 = Maintenance
2 = Service
3 = Production
4 = Development

93Enter access code2177

Integer

Read / Write

0 to 9 999

94

Reset Maintenance code

8880 to 8895

StringRead / Write

Character string comprising numbers,letters and special characters (32)94

"Define Maintenance code" wizard
Navigation: System → User management → Define Maintenance code
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Define Maintenance code

29482

Integer

Read / Write

0 to 9 999

95

Confirm Maintenance code

29481

Integer

Read / Write

0 to 9 999

95

136

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information"Connectivity" submenu"Bluetooth configuration" submenu
Navigation: System → Connectivity → Bluetooth configuration
Parameter

Register

Data type

AccessSelection / User entry / User interface→ Bluetooth

27662

Integer

Read / Write

0 = Disable
1 = Enable
4 = Not available *96

Communication established

27927

IntegerRead

0 = Yes
1 = No96*

Visibility depends on order options or device settings"Date/time" submenu
Navigation: System → Date/time
Parameter

Register

Data type

AccessSelection / User entry / User interface→ Time format

2150

Integer

Read / Write

12 = 12 h AM/PM
24 = 24 h

97Time zone

27339

Integer

Read / Write

0 = UTC 00:00
1 = UTC+01:00
2 = UTC+02:00
3 = UTC+03:00
4 = UTC+04:00
5 = UTC+05:00
6 = UTC+06:00
7 = UTC+07:00
8 = UTC+08:00
9 = UTC+09:00
10 = UTC+10:00
11 = UTC+11:00
12 = UTC+12:00
13 = UTC+13:00
14 = UTC+14:00
35 = UTC+03:30
45 = UTC+04:30
55 = UTC+05:30
57 = UTC+05:45
65 = UTC+06:30
87 = UTC+08:45
95 = UTC+09:30
105 = UTC+10:30
127 = UTC+12:45
135 = UTC-03:30
195 = UTC-09:30
201 = UTC-01:00
202 = UTC-02:00
203 = UTC-03:00
204 = UTC-04:00
205 = UTC-05:00
206 = UTC-06:00
207 = UTC-07:00
208 = UTC-08:00
209 = UTC-09:00
210 = UTC-10:00
211 = UTC-11:00
212 = UTC-12:00

97

Endress+Hauser

137

Modbus RS485 register information

Proline Promag 10 Modbus RS485"Information" submenu"Device" submenu
Navigation: System → Information → Device
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Device name7263 to 7270

StringRead

Character string comprising numbers,letters and special characters

99

Device tag

2026 to 2041

StringRead

Character string comprising numbers,letters and special characters

100

Serial number

7003 to 7008

StringRead

Character string comprising numbers,letters and special characters

100

Order code

2058 to 2067

StringRead

Character string comprising numbers,letters and special characters100

Firmware version

7277 to 7280

StringRead

Character string comprising numbers,letters and special characters

100

Extended order code 1

2212 to 2221

StringRead

Character string comprising numbers,letters and special characters

101

Extended order code 2

2222 to 2231

StringRead

Character string comprising numbers,letters and special characters

101

Extended order code 32232 to 2241

StringRead

Character string comprising numbers,letters and special characters101

ENP version

4003 to 4010

StringRead

Character string comprising numbers,letters and special characters

101

Manufacturer

8001 to 8016

StringRead

Character string comprising numbers,letters and special characters

102"Sensor electronic module (ISEM)" submenu
Navigation: System → Information → Sensor electronic module (ISEM)
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Firmware version

7039

Integer

Read

Positive integer

102"Display module" submenu
Navigation: System → Information → Display module
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Firmware version

7039

Integer

Read

Positive integer

103

138

Endress+Hauser

Proline Promag 10 Modbus RS485

Modbus RS485 register information"Display" submenu
Navigation: System → Display
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Language

35001

Integer

Read / Write

0 = English
1 = Deutsch
2 = Français
3 = Español
4 = Italiano
5 = Nederlands
8 = Svenska
10 = Bahasa Indonesia
11 = 日本語 (Japanese)
12 = Portuguesa
13 = Polski
14 = русский язык (Russian)
15 = čeština (Czech)
16 = 中文 (Chinese)
17 = ภาษาไทย (Thai) *18 = Türkçe
19 = tiếng Việt (Vietnamese)
20 = 한국어 (Korean)
21 = ‫( الْعَرَبيّة‬Arabic) *104

Value 1 display

34918

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3

105

Value 2 display

34919

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3
251 = None

105

Value 3 display

34922

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3
251 = None

106

Value 4 display

34923

Integer

Read / Write

1 = Volume flow
4 = Conductivity *7 = Temperature *9 = Corrected conductivity *11 = Mass flow
16 = Totalizer 1
17 = Totalizer 2
18 = Totalizer 3
251 = None106

Display damping

27602 to 27603

Float

Read / Write

0.0 to 999.9 s

106

Rotation display

36770

Integer

Read / Write

0 = 0 degree
8 = Auto
9 = 90 degree
18 = 180 degree
27 = 270 degree

107

Endress+Hauser

139

Modbus RS485 register information

Proline Promag 10 Modbus RS485

Navigation: System → Display
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Brightness

36768 to 36769

Float

Read / Write

0 to 100 %107

Color scheme

30228

Integer

Read / Write

11 = Light
12 = Dark

107*

Visibility depends on order options or device settings"Software configuration" submenu
Navigation: System → Software configuration
Parameter

Register

Data type

AccessSelection / User entry / User interface→ 

Activate SW option

2795

Integer

Read / Write

Positive integer

108

Software option overview

2902

Integer

Read

16384 = Heartbeat Monitoring
32768 = Heartbeat Verification

108

140

Endress+Hauser

Proline Promag 10 Modbus RS485

Index

Index
A

Activate SW option (Parameter) . . . . . . . . . . . . . . . 108
Active diagnostics (Submenu) . . . . . . . . . . . . . . . . . . 30
Actual diagnostics (Parameter) . . . . . . . . . . . . . . . . . 30
Alarm delay (Parameter) . . . . . . . . . . . . . . . . . . . . . . 39
Application (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 49
Assign behavior of diagnostic no. 043 (Parameter) . . . 40
Assign behavior of diagnostic no. 230 (Parameter) . . . 41
Assign behavior of diagnostic no. 231 (Parameter) . . . 41
Assign behavior of diagnostic no. 302 (Parameter) . . . 41
Assign behavior of diagnostic no. 376 (Parameter) . . . 41
Assign behavior of diagnostic no. 377 (Parameter) . . . 42
Assign behavior of diagnostic no. 441 (Parameter) . . . 43
Assign behavior of diagnostic no. 832 (Parameter) . . . 44
Assign behavior of diagnostic no. 833 (Parameter) . . . 44
Assign behavior of diagnostic no. 834 (Parameter) . . . 45
Assign behavior of diagnostic no. 835 (Parameter) . . . 45
Assign behavior of diagnostic no. 842 (Parameter) . . . 46
Assign behavior of diagnostic no. 937 (Parameter) . . . 46
Assign behavior of diagnostic no. 938 (Parameter) . . . 47
Assign behavior of diagnostic no. 961 (Parameter) . . . 47
Assign behavior of diagnostic no. 962 (Parameter) . . . 48
Assign process variable (Parameter) . . . . . . . . . . . 13, 59
Assign simulation process variable (Parameter) . . . . . 35

BBluetooth (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 96
Bluetooth configuration (Submenu) . . . . . . . . . . . . . . 96
Brightness (Parameter) . . . . . . . . . . . . . . . . . . . . . . 107

C

Calibration (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 70
Calibration factor (Parameter) . . . . . . . . . . . . . . . . . . 71
Color scheme (Parameter) . . . . . . . . . . . . . . . . . . . . 107
Commissioning (Wizard). . . . . . . . . . . . . . . 6, 8, 9, 13, 15, 16, 17, 19, 20, 25, 27
Communication (Submenu) . . . . . . . . . . . . . . . . . . . . 85
Communication established (Parameter) . . . . . . . . . . 96
Conductivity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 50
Conductivity calibration factor (Parameter) . . . . . . . . 71
Conductivity damping time (Parameter) . . . . . . . . . . . 65
Conductivity measurement (Parameter) . . . . . . . . . . . 64
Conductivity temperature coefficient (Parameter) . . . . 65
Conductivity unit (Parameter) . . . . . . . . . . . . . . . 12, 56
Configuration (Submenu) . . . . . . . . . . . . . . . . . . . . . 43
Configuration counter (Parameter) . . . . . . . . . . . . . . 92
Connectivity (Submenu) . . . . . . . . . . . . . . . . . . . . . . 96
Control Totalizer 1 to n (Parameter) . . . . . . . . . . . . . . 60
Corrected conductivity (Parameter) . . . . . . . . . . . . . . 50
Current output 1 (Submenu) . . . . . . . . . . . . . . . . . . . 74
Current output 1 simulation (Parameter) . . . . . . . . . . 36
Current output value (Parameter) . . . . . . . . . . . . . . . 36
Current range output (Parameter) . . . . . . . . . . . . 20, 79

D

Date / Time (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 97
Define access code (Wizard) . . . . . . . . . . . . . . . . . . . 95
Density unit (Parameter) . . . . . . . . . . . . . . . . . . . 12, 56
Device (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Device alarm simulation (Parameter) . . . . . . . . . . . . . 37
Device management (Submenu) . . . . . . . . . . . . . . . . . 91
Device name (Parameter) . . . . . . . . . . . . . . . . . . . . 8, 99
Device reset (Parameter) . . . . . . . . . . . . . . . . . . . . . . 92
Device tag (Parameter) . . . . . . . . . . . . . . . . . 8, 91, 100
Diagnostic configuration (Submenu) . . . . . . . . . . . . . 39
Diagnostic event category (Parameter) . . . . . . . . . . . . 37
Diagnostic event simulation (Parameter) . . . . . . . . . . 37
Diagnostic list (Submenu) . . . . . . . . . . . . . . . . . . . . . 32
Diagnostic settings (Submenu) . . . . . . . . . . . . . . . . . . 39
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Diagnostics 1 (Parameter) . . . . . . . . . . . . . . . . . . . . . 32
Diagnostics 2 (Parameter) . . . . . . . . . . . . . . . . . . . . . 33
Diagnostics 3 (Parameter) . . . . . . . . . . . . . . . . . . . . . 33
Diagnostics 4 (Parameter) . . . . . . . . . . . . . . . . . . . . . 33
Diagnostics 5 (Parameter) . . . . . . . . . . . . . . . . . . . . . 34
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 104
Display damping (Parameter) . . . . . . . . . . . . . . . 26, 106
Display module (Submenu) . . . . . . . . . . . . . . . . . . . 102
Document
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

E

ECC duration (Parameter) . . . . . . . . . . . . . . . . . . . . . 72
ECC interval (Parameter) . . . . . . . . . . . . . . . . . . . . . . 73
ECC polarity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 73
ECC recovery time (Parameter) . . . . . . . . . . . . . . . . . 72
Electrode cleaning cycle (Parameter) . . . . . . . . . . . . . 72
Electrode cleaning cycle (Submenu) . . . . . . . . . . . . . . 72
Electronics (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 40
Empty pipe adjust value (Parameter) . . . . . . . . . . . . . 19
Empty pipe detection (Parameter) . . . . . . . . . . . . . . . 19
ENP version (Parameter) . . . . . . . . . . . . . . . . . . . . . 101
Enter access code (Parameter) . . . . . . . . . . . . . . . . . . 94
Extended order code 1 (Parameter) . . . . . . . . . . . . . 101
Extended order code 2 (Parameter) . . . . . . . . . . . . . 101
Extended order code 3 (Parameter) . . . . . . . . . . . . . 101

F

Failure behavior current output (Parameter) . . . . . 23, 83
Failure current (Parameter) . . . . . . . . . . . . . . . . . 24, 83
Failure mode (Parameter) . . . . . . . . . . . . . . . . . . 15, 61
Firmware version (Parameter) . . . . . . . 8, 100, 102, 103
Fixed current (Parameter) . . . . . . . . . . . . . . . . . . 23, 80
Fixed density (Parameter) . . . . . . . . . . . . . . . . . . . . . 66
Flow damping (Parameter) . . . . . . . . . . . . . . . . . 15, 64
Flow override (Parameter) . . . . . . . . . . . . . . . . . . . . . 64
Full pipe adjust value (Parameter) . . . . . . . . . . . . . . . 19

G

Guidance (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

Damping current output (Parameter) . . . . . . . . . . 23, 82
Endress+Hauser

141

IndexH

Heartbeat Technology (Submenu) . . . . . . . . . . . . . . . 38

I

Information (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 99
Installation direction (Parameter) . . . . . . . . . . . . . . . 69
Integration time (Parameter) . . . . . . . . . . . . . . . . . . . 70

L

Language (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 104
Locking status (Parameter) . . . . . . . . . . . . . . . . . . . . 91
Low flow cut off (Parameter) . . . . . . . . . . . . . . . . 17, 66
Low flow cut off (Submenu) . . . . . . . . . . . . . . . . . . . . 66
Lower range value output (Parameter) . . . . . . . . . 21, 80

M

Manufacturer (Parameter) . . . . . . . . . . . . . . . . . . . . 102
Mass flow (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 49
Mass flow unit (Parameter) . . . . . . . . . . . . . . . . . 11, 55
Mass unit (Parameter) . . . . . . . . . . . . . . . . . . . . . 11, 55
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 49
Measuring mode current output (Parameter) . . . . . . . 75
Measuring period (Parameter) . . . . . . . . . . . . . . . . . . 70
Menu
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Guidance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 90
Modbus configuration (Submenu) . . . . . . . . . . . . . . . 85
Modbus data map (Submenu) . . . . . . . . . . . . . . . . . . 88
Modbus information (Submenu) . . . . . . . . . . . . . . . . 89

NNominal diameter (Parameter) . . . . . . . . . . . . . . . . . 70

O

Off value low flow cutoff (Parameter) . . . . . . . . . . 18, 67
On value low flow cutoff (Parameter) . . . . . . . . . . 18, 67
Operating time (Parameter) . . . . . . . . . . . . . . . . . . . . 31
Operating time from restart (Parameter) . . . . . . . . . . 31
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . 100
Output current 1 (Parameter) . . . . . . . . . . . . . . . . . . 84

P

Partially filled pipe detection (Submenu) . . . . . . . . . . 67
Preset value 1 to n (Parameter) . . . . . . . . . . . . . . . . . 61
Pressure shock suppression (Parameter) . . . . . . . . . . 16
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . . 30
Process (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 43
Process parameters (Submenu) . . . . . . . . . . . . . . . . . 63
Process variable current output (Parameter) . . . . . 20, 74
Process variable value (Parameter) . . . . . . . . . . . . . . . 36
Properties (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 39

R

Reset all totalizers (Parameter) . . . . . . . . . . . . . . . . . 58
Reset Maintenance code (Parameter) . . . . . . . . . . . . . 94
Rotation display (Parameter) . . . . . . . . . . . . . . . . . . 107

142

Proline Promag 10 Modbus RS485

S

Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 40, 63
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . . . . 69
Sensor electronic module (ISEM) (Submenu) . . . . . . 102
Serial number (Parameter) . . . . . . . . . . . . . . . . . . 8, 100
Set date/time (Parameter) . . . . . . . . . . . . . . . . . . . . . 28
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 35
Software configuration (Submenu) . . . . . . . . . . . . . . 108
Software option overview (Parameter) . . . . . . . . . . . 108
Submenu
Active diagnostics . . . . . . . . . . . . . . . . . . . . . . . . 30
Bluetooth configuration . . . . . . . . . . . . . . . . . . . . 96
Calibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70
Communication . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Configuration . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43
Connectivity . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 96
Current output 1 . . . . . . . . . . . . . . . . . . . . . . . . . 74
Date / Time . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Device management . . . . . . . . . . . . . . . . . . . . . . . 91
Diagnostic configuration . . . . . . . . . . . . . . . . . . . . 39
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Diagnostic settings . . . . . . . . . . . . . . . . . . . . . . . . 39
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 104
Display module . . . . . . . . . . . . . . . . . . . . . . . . . . 102
Electrode cleaning cycle . . . . . . . . . . . . . . . . . . . . 72
Electronics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Heartbeat Technology . . . . . . . . . . . . . . . . . . . . . 38
Information . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . 66
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 49
Modbus configuration . . . . . . . . . . . . . . . . . . . . . 85
Modbus data map . . . . . . . . . . . . . . . . . . . . . . . . . 88
Modbus information . . . . . . . . . . . . . . . . . . . . . . . 89
Partially filled pipe detection . . . . . . . . . . . . . . . . . 67
Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43
Process parameters . . . . . . . . . . . . . . . . . . . . . . . 63
Properties . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40, 63
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 69
Sensor electronic module (ISEM) . . . . . . . . . . . . . 102
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
Software configuration . . . . . . . . . . . . . . . . . . . . 108
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Totalizer handling . . . . . . . . . . . . . . . . . . . . . . . . 58
Totalizers . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
User management . . . . . . . . . . . . . . . . . . . . . . . . 93
System (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 90
System units (Submenu) . . . . . . . . . . . . . . . . . . . . . . 53

T

Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Temperature (Parameter) . . . . . . . . . . . . . . . . . . . . . 51
Temperature damping time (Parameter) . . . . . . . . . . 66
Temperature unit (Parameter) . . . . . . . . . . . . . . . 12, 56
Time format (Parameter) . . . . . . . . . . . . . . . . . . . . . . 27
Time zone (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 27
Endress+Hauser

Proline Promag 10 Modbus RS485

Index

Timestamp (Parameter) . . . . . . . . . . . 30, 31, 32, 33, 34
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 51
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . . . . 58
Totalizer handling (Submenu) . . . . . . . . . . . . . . . . . . 58
Totalizer operation mode (Parameter) . . . . . . . . . 14, 60
Totalizer overflow 1 to n (Parameter) . . . . . . . . . . . . . 52
Totalizer value 1 to n (Parameter) . . . . . . . . . . . . . . . 51
Totalizers (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 58

U

Unit totalizer 1 to n (Parameter) . . . . . . . . . . . . . 13, 59
Upper range value output (Parameter) . . . . . . . . . 22, 82
User management (Submenu) . . . . . . . . . . . . . . . . . . 93
User role (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 93

V

Value 1 display (Parameter) . . . . . . . . . . . . . . . . 25, 105
Value 2 display (Parameter) . . . . . . . . . . . . . . . . 25, 105
Value 3 display (Parameter) . . . . . . . . . . . . . . . . 26, 106
Value 4 display (Parameter) . . . . . . . . . . . . . . . . 26, 106
Volume flow (Parameter) . . . . . . . . . . . . . . . . . . . . . 50
Volume flow unit (Parameter) . . . . . . . . . . . . . . . . 9, 53
Volume unit (Parameter) . . . . . . . . . . . . . . . . . . . 10, 54

W

WizardCommissioning 6, 8, 9, 13, 15, 16, 17, 19, 20, 25, 27
Define access code . . . . . . . . . . . . . . . . . . . . . . . . 95

Z

Zero point (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 71

Endress+Hauser

143

*71575137*
71575137

www.addresses.endress.com