GP01209D/06/EN/01.24-00
71654754
2024-06-12

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Proline Promag 10
Electromagnetic flowmeter
IO-Link

Services

Proline Promag 10 IO-Link

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

5.5

1.1
1.2
1.3

5.6
5.7

6

Explanation of abbreviated units . . . 90

1.4

Document function . . . . . . . . . . . . . . . . . . . . .
Target group . . . . . . . . . . . . . . . . . . . . . . . . . .
Using this document . . . . . . . . . . . . . . . . . . . .
1.3.1 Symbols . . . . . . . . . . . . . . . . . . . . . . .
1.3.2 Information on the document
structure . . . . . . . . . . . . . . . . . . . . . . .
1.3.3 Operation concept . . . . . . . . . . . . . . . .
1.3.4 Structure of a parameter description . .
Related documentation . . . . . . . . . . . . . . . . . .

"Information" submenu . . . . . . . . . . . . . . . . . . 80
5.5.1 "Device" submenu . . . . . . . . . . . . . . . . 80
5.5.2 "Sensor electronic module (ISEM)"
submenu . . . . . . . . . . . . . . . . . . . . . . 83
5.5.3 "Display module" submenu . . . . . . . . . 83
"Display" submenu . . . . . . . . . . . . . . . . . . . . . 85
"Software configuration" submenu . . . . . . . . . . 89

2

"Guidance" menu . . . . . . . . . . . . . . . . . . . 7

6.1
6.2
6.3

SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 90
US units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 90
Imperial units . . . . . . . . . . . . . . . . . . . . . . . . 91

2.1

"Commissioning" menu . . . . . . . . . . . . . . . . . . . 7
2.1.1 "Device identification" wizard . . . . . . . . 7
2.1.2 "System units" wizard . . . . . . . . . . . . . . 8
2.1.3 Totalizer 1 to n . . . . . . . . . . . . . . . . . 13
2.1.4 "Measuring conditions" wizard . . . . . . 15
2.1.5 "Display" wizard . . . . . . . . . . . . . . . . . 19
2.1.6 "Date/time" wizard . . . . . . . . . . . . . . . 22

3

"Diagnostics" menu . . . . . . . . . . . . . . . . 24

3.1
3.2
3.3
3.4
3.5
3.6

"Active diagnostics" submenu . . . . . . . . . . . . . 25
"Diagnostic list" submenu . . . . . . . . . . . . . . . . 28
"Event logbook" submenu . . . . . . . . . . . . . . . . 32
"Simulation" submenu . . . . . . . . . . . . . . . . . . . 33
"Heartbeat Technology" submenu . . . . . . . . . . 35
"Diagnostic settings" submenu . . . . . . . . . . . . . 36
3.6.1 "Properties" submenu . . . . . . . . . . . . . 36
3.6.2 "Diagnostic configuration" submenu . . 36

4"Application" menu . . . . . . . . . . . . . . . . 45

4.1

4.5

"Measured values" submenu . . . . . . . . . . . . . . 45
4.1.1 "Totalizer" submenu . . . . . . . . . . . . . . 47
"System units" submenu . . . . . . . . . . . . . . . . . 49
"Totalizers" submenu . . . . . . . . . . . . . . . . . . . 53
4.3.1 "Totalizer handling" submenu . . . . . . . 53
4.3.2 "Totalizer 1 to n" submenu . . . . . . . . . 53
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . . 58
4.4.1 "Process parameters" submenu . . . . . . 58
4.4.2 "Low flow cutoff" submenu . . . . . . . . . 61
4.4.3 "Empty pipe detection" submenu . . . . . 62
4.4.4 "Sensor adjustment" submenu . . . . . . . 64
4.4.5 "Calibration" submenu . . . . . . . . . . . . 65
4.4.6 "Electrode cleaning cycle" submenu . . . 66
"IO-Link" submenu . . . . . . . . . . . . . . . . . . . . . 69

5

"System" menu . . . . . . . . . . . . . . . . . . . . 71

5.1
5.2

"Device management" submenu . . . . . . . . . . . . 72
"User management" submenu . . . . . . . . . . . . . 74
5.2.1 "Define Maintenance code" wizard . . . 76
"Connectivity" submenu . . . . . . . . . . . . . . . . . 77
5.3.1 "Bluetooth configuration" submenu . . . 77
" Date / Time" submenu . . . . . . . . . . . . . . . . . 78

4.2
4.3
4.4

5.3
5.4

Endress+Hauser

4
4
4
4
4
5
6
6

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 92

3

About this document

Proline Promag 10 IO-Link

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the operating
menus.
It is used to perform tasks that require detailed knowledge of the function of the device:• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Symbols

Types of information
Preferred procedures, processes or actions
Permitted procedures, processes or actions
Forbidden procedures, processes or actions
Additional information
Reference to documentationA

Reference to page
Reference to graphic

1.3.2

Information on the document structure

The parameters of all the operating menus and the commissioning wizard are described inthis document.
• Guidance menu with the Commissioning wizard (→  7), which guides the user
automatically through all the device parameters that are required for commissioning• Application menu (→  45)• Diagnostics menu (→  24)• System menu (→  71)

4

Endress+Hauser

Proline Promag 10 IO-Link

About this document

1.3.3

Operation concept

Operation method

Operation via:
• SmartBlue app 1)• Commubox FXA291

Reliable operation• Operation in local language• Standardized operating concept on the device and in the SmartBlue app• Write protection• When electronics modules are replaced: configurations are transferred using
the T-DAT Backup device memory. The device memory contains process
data, device data and the event logbook. No reconfiguration is necessary.

Diagnostic behavior

Efficient diagnostic behavior increases measurement availability:• Open troubleshooting measures via local display and SmartBlue app.
• Diverse simulation options• Logbook of events that have occurred.

1)

Optional via order code "Display; operation", options H, J or K

IO-LinkThe device-specific parameters are configured via IO-Link. There are specificconfiguration or operating programs from different manufacturers available to theuser for this purpose. The device description file (IODD) is provided for the device
IO-Link operating concept
Operator-oriented menu structure for user-specific tasks. Efficient diagnostic behavior
increases measurement availability:• Diagnostic messages• Remedial measures• Simulation options
IODD download
Two options for downloading the IODD:• www.endress.com/download• https://ioddfinder.io-link.com/
www.endress.com/download
1.

Select "Device drivers".

2.

Under "Type", select the "IO Device Description (IODD)" item.

3.

Select "Product root".

4.

Click "Search ".
 A list of search results is displayed.

Select and download the appropriate version.
https://ioddfinder.io-link.com/
1.

Enter and select "Endress" as the manufacturer.

2.

Select product name.
 A list of search results is displayed.

Select and download the appropriate version.
For detailed IO-Link information, see "IO-Link" Special Documentation on the device→ 6

Endress+Hauser

5About this document

Proline Promag 10 IO-Link

1.3.4

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter nameWrite-protected parameter =Navigation

Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Selection

List of the individual options for the parameter• Option 1• Option 2

User entry

Input range for the parameter

User interface

Display value/data for the parameter

Additional information

Additional explanations (e.g. in examples):• On individual options• On display values/data• On the input range• On the parameter function

1.4

Related documentation

Technical information

Overview of the device with the most important technical data.

Operating instructions

All the information that is required in the various phases of the life cycle ofthe device: from product identification, incoming acceptance and storage, tomounting, connection, operation and commissioning through totroubleshooting, maintenance and disposal as well as the technical data anddimensions.

Sensor Brief Operating Instructions

Incoming acceptance, transport, storage and mounting of the device.

Transmitter Brief Operating
Instructions

Electrical connection and commissioning of the device.

Description of Parameters

Detailed explanation of the menus and parameters.

Safety Instructions

Documents for the use of the device in hazardous areas.

Special Documentation

Documents with more detailed information on specific topics.

Installation Instructions

Installation of spare parts and accessories.

The related documentation is available online:
Device Viewer

On the www.endress.com/deviceviewer website, enter the serial number ofthe device: nameplate

Endress+Hauser Operations App‣ Scan the Data Matrix code: nameplate‣ Enter the serial number of the device: nameplate

6

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

2

"Guidance" menu

Main functions for use – from fast and safe commissioning to guided support during
operation.
Navigation

 Guidance

Guidance‣ Commissioning

2.1

→ 7

"Commissioning" menu

Complete this wizard to commission the device. NOTE: If you exit the wizard beforehand,the changes you made will be saved. For this reason, the device may be in an undefinedstate! In this case, reset the device to the default settings.
Navigation

 Guidance → Commissioning‣ Commissioning

2.1.1
Navigation‣ Device identification→ 7‣ System units→ 8‣ Totalizer 1 to n

→  13‣ Measuring conditions→  15‣ Display→  19‣ Date/time→  22"Device identification" wizard
 Guidance → Commissioning → Device ident.

‣ Device identification

Endress+Hauser

Device name→ 8

Serial number→ 8

Firmware version

→ 8

7

"Guidance" menu

Proline Promag 10 IO-Link

Device name
Navigation

Guidance → Commissioning → Device ident. → Device name

Description

Displays the name of the transmitter. The transmitter name is also provided on the
nameplate of the transmitter.

User interface

Character string comprising numbers, letters and special characters

Serial number
Navigation

Guidance → Commissioning → Device ident. → Serial number

Description

Displays the serial number of the measuring device. The serial number is also provided on
the nameplate of the sensor and of the transmitter.
The serial number can also be used to retrieve further device-related information anddocumentation via the Operations app or the Device Viewer on the Endress+Hauser
website.

User interface

Character string comprising numbers, letters and special characters

Firmware version
Navigation

Guidance → Commissioning → Device ident. → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

2.1.2"System units" wizard

Navigation

 Guidance → Commissioning → System units‣ System units

8

Volume flow unit

→ 9Mass flow unit

→  11

Density unit

→  11

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

Temperature unit

→  12

Conductivity unit

→  12

Volume flow unitNavigationDescription

Select the volume flow unit.

Endress+Hauser

Guidance → Commissioning → System units → Volume flow unit

9

"Guidance" menu

Proline Promag 10 IO-Link

Selection

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dAdditional information

Options

US units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• MMft³/s• MMft³/min• MMft³/h• Mft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)

For an explanation of the abbreviated units: →  90
The IO-Link interface only offers the m³/h option.

10

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

Mass flow unitNavigationDescription

Select the mass flow unit.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

Additional information

Guidance → Commissioning → System units → Mass flow unit

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

The IO-Link interface only offers the kg/s option.

Density unitNavigation

Guidance → Commissioning → System units → Density unit

Description

Select the density unit.

Selection

SI units• g/cm³• g/m³• kg/l• kg/dm³• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

Additional information

Options

US units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)

For an explanation of the abbreviated units: →  90
The IO-Link interface only offers the kg/m³ option.

Endress+Hauser

11

"Guidance" menu

Proline Promag 10 IO-Link

Temperature unitNavigation

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (enhanced transmitter) and order code for "Sensor
option", option CI (fluid temperature measurement)

Description

Select the temperature unit.

Selection

SI units• °C•K

Additional information

Guidance → Commissioning → System units → Temperature unit

US units• °F• °R

The IO-Link interface only offers the °C option.

Conductivity unitNavigation

Guidance → Commissioning → System units → Conductiv. unit

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (enhanced transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Select the conductivity unit.

Selection

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/m
• kS/m• MS/mAdditional information

12

The IO-Link interface only offers the unit S/m option.

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

2.1.3

Totalizer 1 to n

Navigation

 Guidance → Commissioning → Totalizer 1 to n

‣ Totalizer 1 to n
Assign process variable 1 to n

→  13

Process variable unit 1 to n

→  13

Totalizer 1 to n operation mode→  14

Totalizer 1 to n failure behavior→  15Assign process variable

Navigation

Guidance → Commissioning → Totalizer 1 to n → AssignVariab. 1 to n

Description

Select a process variable to activate the totalizer.
If the process variable is changed or the totalizer deactivated, the totalizer is reset to "0".

Selection• Off• Volume flow
• Mass flow
Totalizer 1 is permanently set toVolume flow option and cannot be changed.
Totalizers 2 and 3 can be changed.

Additional information

Process variable unitNavigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select the unit for the process variable of the totalizer.

Selection

SI units• g*• kg *• t**

Guidance → Commissioning → Totalizer 1 to n → VariableUnit 1 to n

US units• oz *• lb *• STon *Visibility depends on order options or device settingsorEndress+Hauser

13

"Guidance" menu

Proline Promag 10 IO-Link

SI units• cm³ *
• dm³ *
• m³ *• ml *• l*• hl *• Ml Mega **US units• af *• ft³ *• Mft³ *• fl oz (us) *• gal (us) *• kgal (us) *• Mgal (us) *• bbl (us;liq.) *• bbl (us;beer) *• bbl (us;oil) *• bbl (us;tank) *Imperial units• gal (imp) *• Mgal (imp) *• bbl (imp;beer) *• bbl (imp;oil) *Visibility depends on order options or device settingsorOther units
None **

Additional information

Visibility depends on order options or device settings

Description
The unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  8).
Options
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  13).
• The IO-Link interface only offers the kg option, m³ option and Nm³ option.
• Totalizer 1 is permanently set to Volume flow option and cannot be changed. Totalizers
2 and 3 can be changed.

Totalizer operation modeNavigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select the totalizer operation mode, e.g. only totalize forward flow or only totalize reverse
flow.

Selection• Net• Forward• Reverse

14

Guidance → Commissioning → Totalizer 1 to n → Operat. mode 1 to n

Endress+Hauser

Proline Promag 10 IO-Link

Additional information"Guidance" menu

Selection• Net option
The flow values in the forward and reverse flow directions are totalized and netted
against each other. Net flow is recorded in the flow direction.
• Forward option
Only the flow in the forward flow direction is totalized.
• Reverse option
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Totalizer failure behavior

Navigation

Guidance → Commissioning → Totalizer 1 to n → FailureBehav. 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify how the totalizer should behave in the event of a device alarm.

Selection• Hold• Continue• Last valid value + continue

Additional information

Selection• Hold option
The totalizer is stopped in the event of a device alarm.
• Continue option
The totalizer continues to totalize based on the current value measured; the device alarmis ignored.
• Last valid value + continue option
The totalizer continues to totalize based on the last valid value measured before thedevice alarm occurred.

2.1.4
Navigation"Measuring conditions" wizard


Guidance → Commissioning → Meas. conditions‣ Measuring conditions

Endress+Hauser

Flow damping→  16

Low flow cutoff→  16

On value low flow cutoff→  17

Off value low flow cutoff→  17

Pressure shock suppression→  1815

"Guidance" menu

Proline Promag 10 IO-Link

Empty pipe detection→  19Empty pipe adjust value→  19Full pipe adjust value→  19Flow damping

Navigation

Guidance → Commissioning → Meas. conditions → Flow damping

Description

Enter value for damping of the flow measured value in order to reduce the variability of
the flow measured value when exposed to interference.
Additional information:
The depth of the flow filter is determined by this setting. As the filter depth increases, sodoes the reaction time of the device.
- Value = 0: No damping. Damping of 0 is not recommended, as the measuring signal isthen so noisy that it is almost impossible to perform a measurement.
- Value > 0: Damping increases
Optimal damping depends on the measuring period.
Damping impacts the following measuring device variables:- Outputs- Low flow cutoff- Totalizers

User entry

0 to 15

Low flow cutoff

NavigationDescription

Select a process variable for low flow cutoff to activate low flow cutoff.

Selection• Off• Volume flow
• Mass flow

Additional information

Description

16

Guidance → Commissioning → Meas. conditions → Low flow cutoff

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn-value entered
Off-value entered

On value low flow cutoff

Navigation

Guidance → Commissioning → Meas. conditions → On value

Description

Enter on value to switch on low flow cutoff.
Value = 0: No low flow cutoff
Value > 0: Low flow cutoff is activatedUser entry

Positive floating-point number

Off value low flow cutoff

NavigationDescription

Enter off value to switch off low flow cutoff. The off value is entered as a positive
hysteresis with respect to the on value.

User entry

0 to 100.0 %Endress+Hauser

Guidance → Commissioning → Meas. conditions → Off value

17

"Guidance" menu

Proline Promag 10 IO-Link

Pressure shock suppression

Navigation

Guidance → Commissioning → Meas. conditions → Pres. shock sup.

Description

Enter a time span for signal suppression (= pressure shock suppression active), forexample to prevent the device from registering flow movements in the pipe when a valve isclosed.
Pressure schock suppression is activated when the flow rate drops below the on value forlow flow cutoff.
Values reported when pressure shock suppression is active:
Flow: 0Totalizer: Last valid value
Pressure shock suppression is deactivated when the time span specified has elapsed andthe flow rate exceeds the off value for low flow cutoff.

User entry

0 to 100 s

Additional information

ExampleWhen a valve is closed, momentarily strong fluid movements may occur in the pipeline,
which are registered by the device. These totalized flow values lead to a false totalizer
status, particularly during batching processes.
C

D

Q

2

1

D

3

6
5

4

t

A

B
A0012888

Q
t
A
B
C
D
1
2
3
4
5
6

18

Flow
Time
After runPressure shock
Pressure shock suppression active according to the time entered
Pressure shock suppression inactive
Valve closes
Flow falls below the on-value of the low flow cut off: pressure shock suppression is activated
The time entered has elapsed: pressure shock suppression is deactivated
The current flow value is processed and displayed again.
On value for low flow cut off
Off value for low flow cut off

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

Empty pipe detection



Navigation

Guidance → Commissioning → Meas. conditions → Empty pipe det.

Description

Switch empty pipe detection on or off. Switch on empty pipe detection to detect a partiallyfilled or empty measuring tube.

Selection

• Off
• On

Empty pipe adjust value

Navigation

Guidance → Commissioning → Meas. conditions → Empty pipe value

Description

Displays adjustment value when the measuring tube is empty.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

Full pipe adjust value

Navigation

Guidance → Commissioning → Meas. conditions → Full pipe value

Description

Displays adjustment value when the measuring tube is full.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

2.1.5
Navigation"Display" wizard
 Guidance → Commissioning → Display‣ Display

Endress+Hauser

Value 1 display→  20

Value 2 display→  20

Value 3 display→  2119

"Guidance" menu

Proline Promag 10 IO-Link

Value 4 display→  21

Display damping→  22

Value 1 display

Navigation

Guidance → Commissioning → Display → Value 1 display

Description

Select the measured value to display in the first position on the local display.
The unit is set in the "System units" menu.

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Value 2 display

Navigation

Guidance → Commissioning → Display → Value 2 display

Description

Select the measured value to display in the second position on the local display.
The unit is set in the "System units" menu.

Selection*

20• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 IO-Link"Guidance" menu

Value 3 display

Navigation

Guidance → Commissioning → Display → Value 3 display

Description

Select the measured value to display in the third position on the local display.
The unit is set in the "System units" menu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Value 4 display

Navigation

Guidance → Commissioning → Display → Value 4 display

Description

Select the measured value to display in the fourth position on the local display.
The unit is set in the "System units" menu.

Selection*• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Visibility depends on order options or device settings

Endress+Hauser

21

"Guidance" menu

Proline Promag 10 IO-Link

Display damping

Navigation

Guidance → Commissioning → Display → Display damping

Description

Enter a time constant to set the reaction time of the display to fluctuations in themeasured value (PT1 element).
The smaller the time constant, the faster the display reacts to fluctuations in the measured
value.
If the time constant is set to 0, damping is deactivated.

User entry

0.0 to 999.9 s

2.1.6"Date/time" wizard

Navigation

Guidance → Commissioning → Date/time‣ Date/timeTime format→  22Time zone→  22Set date/time→  23Time format



Navigation

Guidance → Commissioning → Date/time → Time format

Description

Select the time format.

Selection• 24 h• 12 h AM/PMTime zoneNavigationDescription

Select the time zone. Every time the time zone is changed, a logbook entry is created.

22

Guidance → Commissioning → Date/time → Time zone

Endress+Hauser

Proline Promag 10 IO-Link

Selection"Guidance" menu

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+14:00

Set date/timeNavigationDescription

Set the date and local time. Every time the date or time is changed, a logbook entry iscreated.

User entry

Date and time

Endress+Hauser

Guidance → Commissioning → Date/time → Set date/time

23

"Diagnostics" menu

Proline Promag 10 IO-Link

3"Diagnostics" menu

Troubleshooting and preventive maintenance – settings for device behavior during process
and device events as well as assistance and measures for diagnostic purposes.
Navigation

 Diagnostics

Diagnostics

24‣ Active diagnostics→  25‣ Diagnostic list

→  28‣ Event logbook→  32‣ Simulation→  33‣ Heartbeat Technology→  35‣ Diagnostic settings→  36Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

3.1

"Active diagnostics" submenu

Navigation

 Diagnostics → Active diagnos.

‣ Active diagnostics
Actual diagnostics→  25

Active diagnostic IO-Link→  25

Timestamp→  26

Previous diagnostics→  26

Last diagnostic IO-Link→  26

Timestamp→  26

Operating time from restart→  26

Operating time→  27

Actual diagnostics
Navigation

Diagnostics → Active diagnos. → Actual diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the currently active diagnostic message.
If there is more than one pending diagnostic event, the message for the diagnostic eventwith the highest priority is displayed.

User interface

Positive integer

Active diagnostic IO-Link
NavigationDescription

Displays the IO-Link event code for the currently active diagnostic message. If there ismore than one pending diagnostic event, the code for the diagnostic message with thehighest priority is displayed.

User interface

0 to 65 535

Endress+Hauser

Diagnostics → Active diagnos. → ActDiag IO-Link

25

"Diagnostics" menu

Proline Promag 10 IO-Link

Timestamp
Navigation

Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp for the currently active diagnostic message.

User interface

Days (d), hours (h), minutes (m), seconds (s)Previous diagnostics
Navigation

Diagnostics → Active diagnos. → Prev.diagnostics

Prerequisite

At least two diagnostic events have already occurred.

Description

Displays the diagnostic message for the last diagnostic event that has ended.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp of the diagnostic message generated for the last diagnostic eventthat has ended.

User interface

Days (d), hours (h), minutes (m), seconds (s)Operating time from restart
Navigation

Diagnostics → Active diagnos. → Time fr. restart

Description

Indicates how long the device has been in operation since the last time the device wasrestarted.

User interface

Days (d), hours (h), minutes (m), seconds (s)Last diagnostic IO-Link
NavigationDescription

Displays the IO-Link event code for the last diagnostic event that has ended.

26

Diagnostics → Active diagnos. → LastDiag IO-Link

Endress+Hauser

Proline Promag 10 IO-Link

User interface"Diagnostics" menu

0 to 65 535

Operating time
NavigationDescription

Indicates how long the device has been in operation.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Diagnostics → Active diagnos. → Operating time27

"Diagnostics" menu

Proline Promag 10 IO-Link

3.2"Diagnostic list" submenu

Navigation

Diagnostics → Diagnostic list

‣ Diagnostic list
Diagnostics 1→  28Diagnostic 1 IO-Link→  29Timestamp→  29Diagnostics 2→  29Diagnostic 2 IO-Link→  29Timestamp→  29Diagnostics 3→  30Diagnostic 3 IO-Link→  30Timestamp→  30Diagnostics 4→  30Diagnostic 4 IO-Link→  31Timestamp→  30Diagnostics 5→  31

Diagnostic 5 IO-Link→  31Timestamp→  31

Diagnostics 1
NavigationDescription

Displays the currently active diagnostic message with the highest priority.

User interface

Positive integer

28

Diagnostics → Diagnostic list → Diagnostics 1

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

Diagnostic 1 IO-Link
Navigation

Diagnostics → Diagnostic list → Diag. 1 IO-Link

Description

Displays the IO-Link event code for the currently active diagnostic message with thehighest priority.

User interface

0 to 65 535

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 2
Navigation

Diagnostics → Diagnostic list → Diagnostics 2

Description

Displays the currently active diagnostic message with the second highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the second highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostic 2 IO-Link
NavigationDescription

Displays the IO-Link event code for the currently active diagnostic message with thesecond highest priority.

User interface

0 to 65 535

Endress+Hauser

Diagnostics → Diagnostic list → Diag. 2 IO-Link

29

"Diagnostics" menu

Proline Promag 10 IO-Link

Diagnostics 3
Navigation

Diagnostics → Diagnostic list → Diagnostics 3

Description

Displays the currently active diagnostic message with the third highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the third highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostic 3 IO-Link
Navigation

Diagnostics → Diagnostic list → Diag. 3 IO-Link

Description

Displays the IO-Link event code for the currently active diagnostic message with the third
highest priority.

User interface

0 to 65 535

Diagnostics 4
Navigation

Diagnostics → Diagnostic list → Diagnostics 4

Description

Displays the currently active diagnostic message with the fourth highest priority.

User interface

Positive integer

Timestamp
NavigationDescription

Displays the timestamp for the diagnostic message with the fourth highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)30

Diagnostics → Diagnostic list → Timestamp

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

Diagnostics 5
Navigation

Diagnostics → Diagnostic list → Diagnostics 5

Description

Displays the currently active diagnostic message with the fifth-highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the fifth highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostic 4 IO-Link
Navigation

Diagnostics → Diagnostic list → Diag. 4 IO-Link

Description

Displays the IO-Link event code for the currently active diagnostic message with the fourthhighest priority.

User interface

0 to 65 535

Diagnostic 5 IO-Link
NavigationDescription

Displays the IO-Link event code for the currently active diagnostic message with the fifth
highest priority.

User interface

0 to 65 535

Endress+Hauser

Diagnostics → Diagnostic list → Diag. 5 IO-Link

31

"Diagnostics" menu

Proline Promag 10 IO-Link

3.3"Event logbook" submenu

Navigation

 Diagnostics → Event logbook‣ Event logbook
Filter options→  32Filter options

Navigation

Diagnostics → Event logbook → Filter options

Description

Select the category of event notification to display in the event list.
Additional information:
The status signals F, C, S and M are categorized in accordance with VDI/VDE 2650 and
NAMUR Recommendation NE 107.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)Clear event listNavigationDescription

Deletes all entries from the events list. Once this function has been executed, the events
list is empty and all the events are deleted.

Selection• Cancel• Clear data

32

Diagnostics → Event logbook → Clear event list

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

3.4

"Simulation" submenu

Navigation

 Diagnostics → Simulation‣ Simulation
Assign simulation process variable→  33

Process value→  33

Device alarm simulation→  34

Diagnostic event simulation→  34

Assign simulation process variable
NavigationDescription

Select a process variable to activate the simulation.

Selection• Off• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *Additional information

Description

Diagnostics → Simulation → Assign proc.var.

The display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

Process value

Navigation

Diagnostics → Simulation → Process value

Description

Enter the process value to simulate.
The unit is set in the "System units" menu.

User entry*

Signed floating-point number

Visibility depends on order options or device settings

Endress+Hauser

33

"Diagnostics" menu

Proline Promag 10 IO-Link

Device alarm simulation



Navigation

Diagnostics → Simulation → Dev. alarm sim.

Description

Switch the device alarm simulation on or off.
While simulation is in progress, a diagnostic message of the Function Check (C) category isdisplayed.

Selection

• Off
• On

Diagnostic event simulation



NavigationDescription

Select the diagnostic event to simulate.

Selection

Off

34

Diagnostics → Simulation → Diagnostic event

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

3.5

"Heartbeat Technology" submenu

The Heartbeat Technology submenu (→  35) is only available with the optional"Heartbeat Verification + Monitoring" application package.
• Order code for: Application package• Option: EB "Heartbeat Verification + Monitoring"
Detailed information and all descriptions of the device parameters of the application
package are available in the "Heartbeat Verification + Monitoring" Special
Documentation
Navigation

 Diagnostics → Heartbeat Techn.

‣ Heartbeat Technology

Endress+Hauser

35

"Diagnostics" menu

Proline Promag 10 IO-Link

3.6"Diagnostic settings" submenu

Navigation

 Diagnostics → Diag. settings‣ Diagnostic settings

3.6.1‣ Properties→  36‣ Diagnostic configuration→  36"Properties" submenu

Navigation

 Diagnostics → Diag. settings → Properties‣ Properties
Alarm delay→  36

Alarm delay

Navigation

Diagnostics → Diag. settings → Properties → Alarm delay

Description

Enter a delay to suppress momentarily pending diagnostic messages.
Only applies to diagnostic events that allow for a delay before the diagnostic message is
generated.

User entry

0 to 60 s

3.6.2
Navigation"Diagnostic configuration" submenu
 Diagnostics → Diag. settings → Diag. config.

‣ Diagnostic configuration

36‣ Sensor→  37‣ Electronics→  37‣ Process→  40Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

"Sensor" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Sensor‣ Sensor
Assign behavior of diagnostic no. 043→  37

Assign behavior of diagnostic no. 043



Navigation

Diagnostics → Diag. settings → Diag. config. → Sensor → Diagnostic no. 043

Description

Select behavior for diagnostic event "043 Sensor short circuit detected".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

"Electronics" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics‣ Electronics

Endress+Hauser

Assign behavior of diagnostic no. 230→  38Assign behavior of diagnostic no. 231→  38Assign behavior of diagnostic no. 302→  3837

"Diagnostics" menu

Proline Promag 10 IO-Link

Assign behavior of diagnostic no. 376→  38Assign behavior of diagnostic no. 377→  39Assign behavior of diagnostic no. 230
NavigationDescription

Select behavior for diagnostic event "230 Date/time incorrect".

Selection• Alarm• Warning• Logbook entry only

Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 230

Assign behavior of diagnostic no. 231
NavigationDescription

Select behavior for diagnostic event "231 Date/time not available".

Selection• Alarm• Warning• Logbook entry only

Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 231

Assign behavior of diagnostic no. 302
NavigationDescription

Select behavior for diagnostic event "302 Device verification active".

Selection• Off• Warning• Logbook entry only

Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 302

Assign behavior of diagnostic no. 376
NavigationDescription

Select behavior for diagnostic event "376 Sensor electronics (ISEM) faulty".

38



Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 376

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 377
NavigationDescription

Select behavior for diagnostic event "377 Electrode signal integrity".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 377• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Endress+Hauser

39

"Diagnostics" menu

Proline Promag 10 IO-Link"Process" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process‣ Process
Assign behavior of diagnostic no. 832→  40Assign behavior of diagnostic no. 833→  41

Assign behavior of diagnostic no. 834→  41

Assign behavior of diagnostic no. 835→  42

Assign behavior of diagnostic no. 842→  42

Assign behavior of diagnostic no. 937→  43

Assign behavior of diagnostic no. 938→  43

Assign behavior of diagnostic no. 961→  44

Assign behavior of diagnostic no. 962→  44

Assign behavior of diagnostic no. 832

Navigation

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 832

Description

Select behavior for diagnostic event "832 Sensor electronics temperature too high".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

40

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

Assign behavior of diagnostic no. 833
NavigationDescription

Select behavior for diagnostic event "833 Sensor electronics temperature too low".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 833• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 834
NavigationDescription

Select behavior for diagnostic event "834 Process temperature too high".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 834• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Endress+Hauser

41

"Diagnostics" menu

Proline Promag 10 IO-Link

Assign behavior of diagnostic no. 835
NavigationDescription

Select behavior for diagnostic event "835 Process temperature too low".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 835• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 842
NavigationDescription

Select behavior for diagnostic event "842 Process value below limit".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 842• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

42

Endress+Hauser

Proline Promag 10 IO-Link"Diagnostics" menu

Assign behavior of diagnostic no. 937
NavigationDescription

Select behavior for diagnostic event "937 Sensor symmetry".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 937• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 938
NavigationDescription

Select behavior for diagnostic event "938 Coil current not stable".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 938• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Endress+Hauser

43

"Diagnostics" menu

Proline Promag 10 IO-Link

Assign behavior of diagnostic no. 961

Navigation

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 961

Description

Select behavior for diagnostic event "961 Electrode potential out of specification".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Assign behavior of diagnostic no. 962
NavigationDescription

Select behavior for diagnostic event "962 Pipe empty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 962• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

44

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

4"Application" menu

Targeted optimization to the application – comprehensive device settings from sensor
technology to system integration for optimum application adaptation.
Navigation

 Application

Application

4.1‣ Measured values→  45‣ System units→  49‣ Totalizers→  53‣ Sensor→  58‣ IO-Link→  69"Measured values" submenu

Navigation

 Application → Measured values‣ Measured values
Mass flow→  45

Volume flow

→  46

Conductivity→  46

Corrected conductivity→  46

Temperature→  47‣ Totalizer→  47Mass flow
Navigation

Application → Measured values → Mass flow

Description

Displays the mass flow calculated.
The unit is set in the "System units" menu.

User interface

Endress+Hauser

Signed floating-point number

45

"Application" menu

Proline Promag 10 IO-Link

Additional information

The IO-Link interface only offers the kg/s option.

Volume flow
Navigation

Application → Measured values → Volume flow

Description

Displays the volume flow measured.
The unit is set in the "System units" menu.

User interface

Signed floating-point number

Additional information

The IO-Link interface only offers the m³/h option.

Conductivity
Navigation

Application → Measured values → Conductivity

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (enhanced transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the conductivity measured.
The unit is set in the "System units" menu.

User interface

Positive floating-point number
The IO-Link interface only offers the S/m option.

Additional information

Corrected conductivity
Navigation

Application → Measured values → CorrConductivity

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the conductivity measured compensated for temperature.
The unit is set in the "System units" menu.

46

Endress+Hauser

Proline Promag 10 IO-Link

User interface"Application" menu

Positive floating-point number

Temperature
Navigation

Application → Measured values → Temperature

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (enhanced transmitter) and order code for "Sensor
option", option CI (fluid temperature measurement)

Description

Displays the medium temperature measured.
The unit is set in the "System units" menu.

User interface

Positive floating-point number

Additional information

The IO-Link interface only offers the °C option.

4.1.1"Totalizer" submenu

Navigation

 Application → Measured values → Totalizer‣ Totalizer
Totalizer 1 to n value→  47

Totalizer 1 to n overflow→  48

Totalizer value
Navigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Endress+Hauser

Application → Measured values → Totalizer → Tot. 1 to n value

47

"Application" menu

Description

Proline Promag 10 IO-Link

Displays the totalizer counter since the last reset.
This parameter can only display figures up to 7 digits. If the counter exceeds this range,
the overflow is displayed in the "Totalizer overflow " parameter.
Example:Value of "Totalizer value" parameter: 1,968,457 m³
Value of "Totalizer overflow " parameter: 1 × 10^7 (1 overflow) = 10,000,000 m³
Counter (total): 11,968,457 m³
In the event of a fault condition, the totalizer behaves as specified in the "Totalizer failure
behavior" parameter.

User interface

Signed floating-point number

Additional information

Totalizer 1 is permanently set to volume flow and cannot be changed. Totalizers 2 and3 can be changed.

Totalizer overflow

Navigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Displays the number of overflows for the totalizer counter ("Totalizer value" parameter).

User interface–32 000.0 to 32 000.0

48

Application → Measured values → Totalizer → Tot. 1 to n overflow

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

4.2

"System units" submenu

Navigation

 Application → System units‣ System units
Volume flow unit

→  49Mass flow unit

→  51

Density unit

→  51

Temperature unit

→  52

Conductivity unit

→  52

Volume flow unitNavigationDescription

Select the volume flow unit.

Endress+Hauser

Application → System units → Volume flow unit

49

"Application" menu

Proline Promag 10 IO-Link

Selection

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dAdditional information

Options

US units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• MMft³/s• MMft³/min• MMft³/h• Mft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)

For an explanation of the abbreviated units: →  90
The IO-Link interface only offers the m³/h option.

50

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

Mass flow unitNavigationDescription

Select the mass flow unit.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

Additional information

Application → System units → Mass flow unit

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

The IO-Link interface only offers the kg/s option.

Density unitNavigation

Application → System units → Density unit

Description

Select the density unit.

Selection

SI units• g/cm³• g/m³• kg/l• kg/dm³• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

Additional information

Options

US units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)

For an explanation of the abbreviated units: →  90
The IO-Link interface only offers the kg/m³ option.

Endress+Hauser

51

"Application" menu

Proline Promag 10 IO-Link

Temperature unitNavigation

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (enhanced transmitter) and order code for "Sensor
option", option CI (fluid temperature measurement)

Description

Select the temperature unit.

Selection

SI units• °C•K

Additional information

Application → System units → Temperature unit

US units• °F• °R

The IO-Link interface only offers the °C option.

Conductivity unitNavigation

Application → System units → Conductiv. unit

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (enhanced transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Select the conductivity unit.

Selection

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/m
• kS/m• MS/mAdditional information

52

The IO-Link interface only offers the unit S/m option.

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

4.3

"Totalizers" submenu

Navigation

 Application → Totalizers‣ Totalizers

4.3.1‣ Totalizer handling→  53‣ Totalizer 1 to n

→  53"Totalizer handling" submenu

Navigation

 Application → Totalizers → Totalizer‣ Totalizer handlingReset all totalizers→  53

Reset all totalizers
Navigation

Application → Totalizers → Totalizer → Reset all tot.

Description

Reset all totalizers to "0" and restart the totalizers. The counter readings are not logged
prior to the reset.

Selection• Cancel• Reset + totalize

4.3.2"Totalizer 1 to n" submenu

Navigation

 Application → Totalizers → Totalizer 1 to n

‣ Totalizer 1 to n

Endress+Hauser

Assign process variable 1 to n

→  54

Process variable unit 1 to n

→  54

Totalizer 1 to n operation mode→  55

Totalizer 1 to n control→  5653

"Application" menu

Proline Promag 10 IO-Link

Preset value 1 to n

→  56

Totalizer 1 to n failure behavior→  57

Assign process variable

Navigation

Application → Totalizers → Totalizer 1 to n → AssignVariab. 1 to n

Description

Select a process variable to activate the totalizer.
If the process variable is changed or the totalizer deactivated, the totalizer is reset to "0".

Selection• Off• Volume flow
• Mass flow

Additional information

Totalizer 1 is permanently set toVolume flow option and cannot be changed.
Totalizers 2 and 3 can be changed.

Process variable unitNavigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select the unit for the process variable of the totalizer.

Selection

SI units• g*• kg *• t**

Application → Totalizers → Totalizer 1 to n → VariableUnit 1 to n

US units• oz *• lb *• STon *Visibility depends on order options or device settingsor54

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

SI units• cm³ *
• dm³ *
• m³ *• ml *• l*• hl *• Ml Mega **US units• af *• ft³ *• Mft³ *• fl oz (us) *• gal (us) *• kgal (us) *• Mgal (us) *• bbl (us;liq.) *• bbl (us;beer) *• bbl (us;oil) *• bbl (us;tank) *Imperial units• gal (imp) *• Mgal (imp) *• bbl (imp;beer) *• bbl (imp;oil) *Visibility depends on order options or device settingsorOther units
None **

Additional information

Visibility depends on order options or device settings

Description
The unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  8).
Options
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  13).
• The IO-Link interface only offers the kg option, m³ option and Nm³ option.
• Totalizer 1 is permanently set to Volume flow option and cannot be changed. Totalizers
2 and 3 can be changed.

Totalizer operation modeNavigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select the totalizer operation mode, e.g. only totalize forward flow or only totalize reverse
flow.

Selection• Net• Forward• Reverse

Endress+Hauser

Application → Totalizers → Totalizer 1 to n → Operat. mode 1 to n

55

"Application" menu

Additional information

Proline Promag 10 IO-Link

Selection• Net option
The flow values in the forward and reverse flow directions are totalized and netted
against each other. Net flow is recorded in the flow direction.
• Forward option
Only the flow in the forward flow direction is totalized.
• Reverse option
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Totalizer control
Navigation

Application → Totalizers → Totalizer 1 to n → Tot. 1 to n control

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Operate the totalizer.

Selection• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• HoldAdditional information

Selection• Totalize option
The totalizer is started or continues running.
• Reset + hold option
The totalizer is reset to "0" and stopped.
• Preset + hold option
The totalizer is stopped and set to the start value specified in the "Preset value "parameter.
• Reset + totalize option
The totalizer is reset to "0" and restarted.
• Hold option
The totalizer is stopped.

Preset value
Navigation

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify a start value for the totalizer.

User entry

Signed floating-point number

56

Application → Totalizers → Totalizer 1 to n → Preset value 1 to n

Endress+Hauser

Proline Promag 10 IO-Link

Additional information"Application" menu

Description
The unit of the selected process variable is specified for the totalizer in the Unit totalizer
parameter (→  13).
ExampleThis configuration is suitable for applications such as iterative filling processes with a fixed
batch quantity.

Totalizer failure behavior

Navigation

Application → Totalizers → Totalizer 1 to n → FailureBehav. 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify how the totalizer should behave in the event of a device alarm.

Selection• Hold• Continue• Last valid value + continue

Additional information

Selection• Hold option
The totalizer is stopped in the event of a device alarm.
• Continue option
The totalizer continues to totalize based on the current value measured; the device alarmis ignored.
• Last valid value + continue option
The totalizer continues to totalize based on the last valid value measured before thedevice alarm occurred.

Endress+Hauser

57

"Application" menu

Proline Promag 10 IO-Link

4.4
Navigation"Sensor" submenu
 Application → Sensor‣ Sensor

4.4.1
Navigation‣ Process parameters→  58‣ Low flow cutoff→  61‣ Empty pipe detection→  62‣ Sensor adjustment→  64‣ Calibration→  65‣ Electrode cleaning cycle→  66"Process parameters" submenu
 Application → Sensor → Process param.

‣ Process parameters

58

Flow damping→  59Flow override→  59Conductivity measurement→  59Conductivity temperature coefficient→  60Conductivity damping time→  60Temperature damping time→  60

Fixed density→  61

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

Flow damping

Navigation

Application → Sensor → Process param. → Flow damping

Description

Enter value for damping of the flow measured value in order to reduce the variability of
the flow measured value when exposed to interference.
Additional information:
The depth of the flow filter is determined by this setting. As the filter depth increases, sodoes the reaction time of the device.
- Value = 0: No damping. Damping of 0 is not recommended, as the measuring signal isthen so noisy that it is almost impossible to perform a measurement.
- Value > 0: Damping increases
Optimal damping depends on the measuring period.
Damping impacts the following measuring device variables:- Outputs- Low flow cutoff- Totalizers

User entry

0 to 15

Flow override

Navigation

Application → Sensor → Process param. → Flow override

Description

Reports the flow rate as zero until flow override is deactivated. Can be used for example
when cleaning the pipeline.

Selection

• Off
• On

Additional information

Selection"On" option
Activates flow override and the diagnostic message "453 Flow override active" is generated.
Values reported:
Flow variables: Zero
Other process variables: As measured
Totalizers: Stop totalizing

Conductivity measurement

Navigation

Prerequisite

Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option", option
CX (conductivity measurement)

Description

Switch conductivity measurement on or off. To be able to measure conductivity, themedium must have a minimum conductivity of 5 μS/cm.

Endress+Hauser

Application → Sensor → Process param. → Conduct. measur.

59

"Application" menu

Selection

Proline Promag 10 IO-Link

• Off
• On

Conductivity temperature coefficientNavigation

Application → Sensor → Process param. → Cond. temp.coeff

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Enter the temperature coefficient to calculate the corrected conductivity.

User entry

Signed floating-point number

Conductivity damping timeNavigation

Application → Sensor → Process param. → ConductDampTime

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Enter time constant for conductivity damping (PT1 element):- Value = 0: No damping- Value > 0: Damping increases
Additional information:
Damping is implemented by means of a proportional transmission behavior with firstorder delay (PT1 element).

User entry

0 to 999.9 s

Temperature damping timeNavigation

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Enter time constant for damping the temperature value.

60

Application → Sensor → Process param. → TempDampingTime

Endress+Hauser

Proline Promag 10 IO-Link

User entry"Application" menu

0 to 999.9 s

Fixed density

Navigation

Application → Sensor → Process param. → Fixed density

Description

Enter a fixed value for the density.
The unit is set in the "" menu.

User entry

Positive floating-point number

4.4.2"Low flow cutoff" submenu

Navigation

 Application → Sensor → Low flow cutoff‣ Low flow cutoff
Low flow cutoff→  61

On value low flow cutoff→  62

Off value low flow cutoff→  62

Low flow cutoff

NavigationDescription

Select a process variable for low flow cutoff to activate low flow cutoff.

Selection• Off• Volume flow
• Mass flow

Additional information

Description

Endress+Hauser

Application → Sensor → Low flow cutoff → Low flow cutoff

61

"Application" menu

Proline Promag 10 IO-Link

Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn-value entered
Off-value entered

On value low flow cutoff

Navigation

Application → Sensor → Low flow cutoff → On value

Description

Enter on value to switch on low flow cutoff.
Value = 0: No low flow cutoff
Value > 0: Low flow cutoff is activatedUser entry

Positive floating-point number

Off value low flow cutoff

Navigation

Application → Sensor → Low flow cutoff → Off value

Description

Enter off value to switch off low flow cutoff. The off value is entered as a positive
hysteresis with respect to the on value.

User entry

0 to 100.0 %4.4.3
Navigation"Empty pipe detection" submenu
 Application → Sensor → Empty pipe det.

‣ Empty pipe detection

62

Empty pipe detection→  63

New adjustment→  63Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

Empty pipe adjust value→  63Full pipe adjust value→  64

Measured value EPD→  64

Empty pipe detection



Navigation

Application → Sensor → Empty pipe det. → Empty pipe det.

Description

Switch empty pipe detection on or off. Switch on empty pipe detection to detect a partiallyfilled or empty measuring tube.

Selection

• Off
• On

New adjustment

Navigation

Application → Sensor → Empty pipe det. → New adjustment

Description

Select empty pipe or full pipe adjustment to perform a new adjustment. To adjust empty
pipe detection, perform the empty pipe adjustment first and then the full pipe adjustment.
Additional information:
The measuring device is pre-adjusted at production using water (approx. 300 μS/cm). For
liquids that deviate from this conductivity, a new empty pipe and full pipe adjustment mustbe performed on site.

Selection• Cancel• Empty pipe adjust• Full pipe adjust

Empty pipe adjust value

Navigation

Application → Sensor → Empty pipe det. → Empty pipe value

Description

Displays adjustment value when the measuring tube is empty.
NOTE
Users logged on in the Service role have write access!

User interface

Endress+Hauser

Positive floating-point number

63

"Application" menu

Proline Promag 10 IO-Link

Full pipe adjust value

Navigation

Application → Sensor → Empty pipe det. → Full pipe value

Description

Displays adjustment value when the measuring tube is full.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

Measured value EPD
Navigation

Application → Sensor → Empty pipe det. → Meas. value EPD

Description

Displays the value currently measured for empty pipe detection.

User interface

Positive floating-point number

4.4.4"Sensor adjustment" submenu

Navigation

 Application → Sensor → Sensor adjustm.

‣ Sensor adjustment
Installation direction→  64Integration time→  65

Measuring period→  65

Installation direction



NavigationDescription

Select the sign of the flow direction.

Selection• Forward flow
• Reverse flow

64

Application → Sensor → Sensor adjustm. → Install. direct.

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

Integration timeNavigation

Application → Sensor → Sensor adjustm. → Integration time

Description

Displays the duration of an integration cycle.
Users logged on in the Service role have write access.

User interface

1 to 65 ms

Measuring period

Navigation

Application → Sensor → Sensor adjustm. → Measuring period

Description

Displays the duration of a full measuring period.
The measuring period is the time span over which a magnetic field is produced to create ameasuring point.
Users logged on in the Service role have write access.

User interface

0 to 1 000 ms

4.4.5"Calibration" submenu

Navigation

 Application → Sensor → Calibration‣ Calibration
Nominal diameter→  65

Calibration factor→  66

Zero point→  66

Conductivity calibration factor→  66

Nominal diameter
NavigationDescription

Displays the nominal diameter of the sensor.

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

Application → Sensor → Calibration → Nominal diameter

65

"Application" menu

Proline Promag 10 IO-Link

Calibration factor
Navigation

Application → Sensor → Calibration → Cal. factor

Description

Displays the current calibration factor for the sensor. The factory setting for the calibration
factor can be found on the sensor's nameplate.

User interface

Positive floating-point number

Zero pointNavigation

Application → Sensor → Calibration → Zero point

Description

Displays the zero point correction value for the sensor.
Users logged on in the Service role have write access.

User interface

Signed floating-point number

Conductivity calibration factor



Navigation

Application → Sensor → Calibration → Cond. cal. fact.

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  59).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the calibration factor for conductivity measurement.
Users logged on in the Service role have write access.

User interface

0.01 to 10 000

4.4.6
Navigation"Electrode cleaning cycle" submenu
 Application → Sensor → Elec. clean cycl

‣ Electrode cleaning cycle

66

Electrode cleaning cycle→  67

ECC duration→  67

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

ECC recovery time→  67

ECC interval→  67

ECC polarity→  68

Electrode cleaning cycleNavigation

Application → Sensor → Elec. clean cycl → Elec. clean cycl

Description

Switch electrode cleaning on or off.

Selection

• Off
• On

ECC duration



Navigation

Application → Sensor → Elec. clean cycl → ECC duration

Description

Specify the duration of the cleaning phase for the cycle. Diagnostic message "530 Electrodecleaning active" is displayed until the cleaning phase and recovery phase are bothcomplete.

User entry

0.01 to 30 s

ECC recovery timeNavigation

Application → Sensor → Elec. clean cycl → ECC recov. time

Description

Specify the maximum timespan for recovery after the cleaning phase has completed toprevent interference with the signal outputs. The output signal values will be frozen forthe duration of the recovery, unless flow measurement can resume beforehand. If thetimespan specified is not sufficient for recovery, diagnostic message "512 ECC recovery
time exceeded" is generated.

User entry

1 to 600 s

ECC interval

NavigationDescription

Specify the duration of the interval between one cleaning cycle and the next.

Endress+Hauser

Application → Sensor → Elec. clean cycl → ECC interval

67

"Application" menu

User entry

Proline Promag 10 IO-Link

0.5 to 168 h

ECC polarity
Navigation

Application → Sensor → Elec. clean cycl → ECC polarity

Description

Displays the setting for the electrode cleaning polarity. The polarity depends on thematerial of the electrodes.

User interface• Positive• Negative

Additional information

User interface• Positive option
For tantalum, Alloy C22, or stainless steel electrodes• Negative option
For platinum electrodes

68

Endress+Hauser

Proline Promag 10 IO-Link"Application" menu

4.5

"IO-Link" submenu

Navigation

 Application → IO-Link‣ IO-Link
Vendor name→  69Product name→  69Product ID→  70Device ID→  69Application specific tag→  70Function tag→  70Location tag→  70Vendor name
Navigation

Application → IO-Link → Vendor name

Description

Displays the manufacturer.

User interface

Character string comprising numbers, letters and special characters

Product name
Navigation

Application → IO-Link → Product name

Description

Displays the name of the transmitter.

User interface

Character string comprising numbers, letters and special characters

Device ID
NavigationDescription

Displays the device ID registered with the IO-Link Community.

User interface

Positive integer

Endress+Hauser

Application → IO-Link → Device ID

69"Application" menu

Proline Promag 10 IO-Link

Product ID
Navigation

Application → IO-Link → Product ID

Description

Displays the product root.

User interface

Character string comprising numbers, letters and special characters

Application specific tag

Navigation

Application → IO-Link → Application tag

Description

Enter the tag of the application in which the device is used, e.g. the designation of the
production process or step (max. 32 characters).

User entry

Character string comprising numbers, letters and special characters (32)Function tag

Navigation

Application → IO-Link → Function tag

Description

Enter the tag of the function the device performs in the application (max. 32 characters).

User entry

Character string comprising numbers, letters and special characters (32)

Location tag

NavigationDescription

Enter the tag of the device location in the plant (max. 32 characters).

User entry

Character string comprising numbers, letters and special characters (32)70

Application → IO-Link → Location tag

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

5

"System" menu

Overall device management and security settings – management of system settings andadaption to operational requirements.
Navigation

 System

SystemEndress+Hauser‣ Device management→  72‣ User management→  74‣ Connectivity→  77‣ Date/time→  78‣ Information→  80‣ Display→  85‣ Software configuration→  8971

"System" menu

Proline Promag 10 IO-Link

5.1"Device management" submenu

Navigation

 System → Device manag.

‣ Device management
Device tag→  72

Locking status→  72

Configuration counter→  73

Device reset→  73

Device tagNavigation



System → Device manag. → Device tag

Description

Displays the name for the measuring point.

User entry

Character string comprising numbers, letters and special characters (32)

Locking status
NavigationDescription

Indicates the write protection with the highest priority that is currently active.

User interface• Hardware locked• Temporarily locked option (e.g. during IO-Link block configuration or parameter
upload)Additional information

72

System → Device manag. → Locking status• The DIP switch is on the back of the display.
• While a block parameterization or the DataStorage mechanism is active via the IOLink communication, the Temporarily locked option becomes active.

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

Configuration counter
Navigation



System → Device manag. → Config. counter

Description

Displays the counter for the number of times the device configuration has changed.
If the value for a static parameter changes, the counter increments by 1. This is to enable
tracking different parameter versions.
When multiple parameters are changed simultaneously, e.g. when loading a configuration
file into the device from an external source such as FieldCare, the counter may increment.
The counter cannot be reset. Nor is it reset to a default value on performing a device reset.
Once the counter has incremented to 65535, it restarts at 1.

User interface

0 to 65 535

Device reset

Navigation



System → Device manag. → Device reset

Description

Reset the device configuration - either entirely or in part - to a defined state.

Selection• Cancel• To delivery settings• Restart device• Restore S-DAT backup *• Create T-DAT backup• Restore T-DAT backup *Additional information

Selection• To delivery settings option
Every parameter for which a customer-specific default setting was ordered is reset to thecustomer-specific value. All other parameters are reset to the factory setting.
• Restart device option
The restart resets every parameter with data stored in volatile memory (RAM) to thefactory setting (e.g. measured value data). The device configuration remains unchanged.
• Restore S-DAT backup option
Restores the data that is saved on the S-DAT.
This function can be used to resolve the memory issue "083 Memory content
inconsistent" or to restore the S-DAT data when a new S-DAT has been installed.
• Create T-DAT backup option
Creates T-DAT backup.
• Restore T-DAT backup option
Restores the data saved on the T-DAT.
This function can be used to resolve the memory issue "283 Memory content
inconsistent" or to restore the T-DAT data when a new T-DAT has been installed.

*

Visibility depends on order options or device settings

Endress+Hauser

73

"System" menu

Proline Promag 10 IO-Link

5.2"User management" submenu

Navigation

 System → User manag.

‣ User management
User role→  74Enter access code→  75

Reset Maintenance code→  75‣ Define Maintenance code→  76

User role
Navigation



System → User manag. → User role

Description

Displays the role the user is currently logged on in.
The role determines the user's access rights for the parameters. Until a "Maintenance" code
has been set in the "Define Maintenance code" parameter, all users are automaticallylogged on in the "Maintenance" role. Once the "Maintenance" code has been set, all users
are automatically logged on in the "Operator" role. The access rights can be changed via the"Enter access code" parameter.

User interface• Operator• Maintenance• Service• Production• Development

Additional information

User interface• Operator option
Provides only read access to parameters.
• Maintenance option
Provides read and write access to parameters.
For some parameters, the user must be logged on in the Service role to obtain write
access.
• Service option
Provides read and write access to Service parameters.

74

Endress+Hauser

Proline Promag 10 IO-Link"System" menuEnter access code
Navigation



System → User manag. → Ent. access code

Description

For users logged on in the Operator role, enter the Maintenance code to change the accessstatus to Maintenance and disable write protection of parameters. For users logged on in
the Maintenance role, enter the Service code to change the access status to Service andenable read and write access to Service parameters.

User entry

Max. 16-digit character string comprising numbers, letters and special characters

Reset Maintenance code
NavigationDescription

Enter the code provided by Endress+Hauser Technical Support to reset the Maintenancecode.

User entry

Character string comprising numbers, letters and special characters (32)Endress+Hauser

System → User manag. → Reset Maint code75

"System" menu

Proline Promag 10 IO-Link

5.2.1"Define Maintenance code" wizard

Complete this wizard to specify an access code for the Maintenance role.
Navigation



System → User manag. → Def. Maint. code

‣ Define Maintenance code
Define Maintenance code→  76

Confirm Maintenance code→  76

Define Maintenance code

Navigation



System → User manag. → Def. Maint. code → Def. Maint. code

Description

Specify an access code that is required to obtain the access rights for the Maintenance role.

User entry

0 to 9 999

Confirm Maintenance code

NavigationDescription

Confirm the access code entered for the Maintenance role.

User entry

0 to 9 999

76

System → User manag. → Def. Maint. code → Conf. Maint code

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

5.3

"Connectivity" submenu

Navigation

 System → Connectivity‣ Connectivity‣ Bluetooth configuration

5.3.1→  77"Bluetooth configuration" submenu

Navigation

 System → Connectivity → Bluetooth conf.

‣ Bluetooth configuration
Bluetooth→  77

Communication established→  77

BluetoothNavigation



System → Connectivity → Bluetooth conf. → Bluetooth

Description

Enable or disable Bluetooth.

Selection• Enable• Disable• Not available *Communication established
Navigation

User interface

• No
• Yes

*

System → Connectivity → Bluetooth conf. → Communi. establ.

Visibility depends on order options or device settings

Endress+Hauser

77

"System" menu

Proline Promag 10 IO-Link

5.4" Date / Time" submenu

Navigation

 System → Date / Time‣ Date/time
Set date/time→  78Time format→  78Time zone→  78Set date/timeNavigation



System → Date/time → Set date/time

Description

Set the date and local time. Every time the date or time is changed, a logbook entry iscreated.

User entry

Date and time

Time format



Navigation



System → Date/time → Time format

Description

Select the time format.

Selection• 24 h• 12 h AM/PMTime zoneNavigationDescription

Select the time zone. Every time the time zone is changed, a logbook entry is created.

78

System → Date/time → Time zone

Endress+Hauser

Proline Promag 10 IO-Link

Selection

Endress+Hauser"System" menu

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+14:00

79

"System" menu

Proline Promag 10 IO-Link

5.5"Information" submenu

Navigation

 System → Information‣ Information

5.5.1‣ Device→  80‣ Sensor electronic module (ISEM)→  83‣ Display module→  83"Device" submenu

Navigation

 System → Information → Device‣ Device
Device name→  80Device tag→  81

Serial number→  81

Order code→  81

Firmware version→  81

Extended order code 1→  82

Extended order code 2→  82

Extended order code 3→  82

ENP version→  82

Manufacturer→  83

Device name
NavigationDescription

Displays the name of the transmitter. The transmitter name is also provided on the
nameplate of the transmitter.

User interface

Character string comprising numbers, letters and special characters

80

System → Information → Device → Device name

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

Device tagNavigation



System → Information → Device → Device tag

Description

Displays the name for the measuring point.

User entry

Character string comprising numbers, letters and special characters (32)Serial number
Navigation



System → Information → Device → Serial number

Description

Displays the serial number of the measuring device. The serial number is also provided on
the nameplate of the sensor and of the transmitter.
The serial number can also be used to retrieve further device-related information anddocumentation via the Operations app or the Device Viewer on the Endress+Hauser
website.

User interface

Character string comprising numbers, letters and special characters

Order code

Navigation



System → Information → Device → Order code

Description

Displays the device order code.
The order code is used for instance to order a replacement or spare device or to verify thatthe device features specified on the order form match the shipping note.

User interface

Character string comprising numbers, letters and special characters

Firmware version
NavigationDescription

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

System → Information → Device → Firmware version

81

"System" menu

Proline Promag 10 IO-Link

Extended order code 1Navigation



System → Information → Device → Ext. order cd. 1

Description

Displays the first, second and/or third part of the extended order code.
Due to character length restrictions, the extended order code is split into a maximum of 3
parameters. The extended order code indicates for each feature in the product structure
the selected option, thereby uniquely identifying the device model.
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Extended order code 2Navigation



System → Information → Device → Ext. order cd. 2

Description

Displays the first, second and/or third part of the extended order code.
Due to character length restrictions, the extended order code is split into a maximum of 3
parameters. The extended order code indicates for each feature in the product structure
the selected option, thereby uniquely identifying the device model.
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Extended order code 3

Navigation



System → Information → Device → Ext. order cd. 3

Description

Displays the first, second and/or third part of the extended order code.
Due to character length restrictions, the extended order code is split into a maximum of 3
parameters. The extended order code indicates for each feature in the product structure
the selected option, thereby uniquely identifying the device model.
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

ENP version
NavigationDescription

Displays the version of the electronic nameplate (ENP).

User interface

Character string comprising numbers, letters and special characters

82

System → Information → Device → ENP version

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

Manufacturer
Navigation



System → Information → Device → Manufacturer

Description

Displays the manufacturer.

User interface

Character string comprising numbers, letters and special characters

5.5.2"Sensor electronic module (ISEM)" submenu

Navigation

 System → Information → Sens. electronic‣ Sensor electronic module (ISEM)
Firmware version→  83

Firmware version
Navigation



System → Information → Sens. electronic → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

5.5.3"Display module" submenu

Navigation

 System → Information → Display module‣ Display module
Firmware version→  83

Firmware version
NavigationDescription

Displays the firmware version of the module.

Endress+Hauser

System → Information → Display module → Firmware version

83

"System" menu

User interface

84

Proline Promag 10 IO-Link

Positive integer

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

5.6

"Display" submenu

Navigation

 System → Display‣ Display
Language→  85

Value 1 display→  86

Value 2 display→  86

Value 3 display→  87

Value 4 display→  87

Display damping→  88

Rotation display→  88Brightness→  88

Color scheme→  88

Language
NavigationDescription

Set display language.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• ‫( الْعَرَبيّة‬Arabic) *• Bahasa Indonesia **System → Display → Language

Visibility depends on order options or device settings

Endress+Hauser

85

"System" menu

Proline Promag 10 IO-Link• ภาษาไทย (Thai) *• tiếng Việt (Vietnamese) *• čeština (Czech)

Value 1 display

Navigation



System → Display → Value 1 display

Description

Select the measured value to display in the first position on the local display.
The unit is set in the "System units" menu.

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Value 2 display

Navigation



System → Display → Value 2 display

Description

Select the measured value to display in the second position on the local display.
The unit is set in the "System units" menu.

Selection*86• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

Value 3 display

Navigation



System → Display → Value 3 display

Description

Select the measured value to display in the third position on the local display.
The unit is set in the "System units" menu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Value 4 display

Navigation



System → Display → Value 4 display

Description

Select the measured value to display in the fourth position on the local display.
The unit is set in the "System units" menu.

Selection*• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3• Noise *• Coil current shot time *Visibility depends on order options or device settings

Endress+Hauser

87

"System" menu

Proline Promag 10 IO-Link

Display damping

Navigation



System → Display → Display damping

Description

Enter a time constant to set the reaction time of the display to fluctuations in themeasured value (PT1 element).
The smaller the time constant, the faster the display reacts to fluctuations in the measured
value.
If the time constant is set to 0, damping is deactivated.

User entry

0.0 to 999.9 s

Rotation display

Navigation



System → Display → Rotation display

Description

Select rotation angle of the display text to optimize local display readability.

Selection• Auto• 0 degree• 90 degree• 180 degree• 270 degree

Brightness
Navigation



System → Display → Brightness

Description

Adjust brightness.

User entry

0 to 100 %Color scheme

NavigationDescription

Select the preferred color scheme.

Selection• Light• Dark88

System → Display → Color scheme

Endress+Hauser

Proline Promag 10 IO-Link"System" menu

5.7"Software configuration" submenu

Navigation

 System → Software config.

‣ Software configuration
Activate SW option→  89Software option overview→  89Activate SW option



Navigation



System → Software config. → Activate SW opt.

Description

Enter application package code or code of the functionality ordered separately to activate
it.
Additional information:- If a measuring device was ordered with an add-on software option, the activation code isprogrammed into the measuring device ex factory.
- After entering the activation code: Check whether the new software option is displayedin the "Software option overview" parameter and therefore active.
NOTE
If an invalid code is entered the software options that have already been activated areinvalidated!
Before entering a new activation code: Create a record of the existing activation code.

User entry

Positive integer

Software option overview
Navigation



System → Software config. → SW option overv.

Description

Displays all software options included in the order ex factory or ordered at a later date thathave been enabled via the operating interface.
If a new software option is not displayed after entering the activation code, the code
entered was inaccurate or invalid. In this case, contact the appropriate Endress+Hauser
sales organization for assistance.

User interface

Endress+Hauser• Extended HistoROM• Heartbeat Verification• Heartbeat Monitoring89

Explanation of abbreviated units

6

Explanation of abbreviated units

6.1

SI units

Process variable Units

Explanation

Densityg/cm³, g/m³

Gram/volume unit

kg/dm³, kg/l, kg/m³

Kilogram/volume unit

SD4°C, SD15°C, SD20°C

Specific density: The specific density is the ratio of thedensity of the fluid to the density of water at a watertemperature of 4 °C (39 °F), 15 °C (59 °F), 20 °C (68 °F).

SG4°C, SG15°C, SG20°C

Specific gravity: The specific gravity is the ratio of thedensity of the fluid to the density of water at a watertemperature of 4 °C (39 °F), 15 °C (59 °F), 20 °C (68 °F).

Mass

g, kg, tGram, kilogram, metric ton

Mass flow

g/s, g/min, g/h, g/dGram/time unit

kg/s, kg/min, kg/h, kg/d

Kilogram/time unit

t/s, t/min, t/h, t/dMetric ton/time unit

Temperature°C , K

Celsius, Kelvin

Volumecm³, dm³, m³

Cubic centimeter, cubic decimeter, cubic meter

ml, l, hl, Ml Mega

Milliliter, liter, hectoliter, megaliterTimes, m, h, d, ySecond, minute, hour, day, year

6.2

US units

Process variable Units

Explanation

Densitylb/ft³, lb/gal (us)Pound/cubic foot, pound/gallon

lb/bbl (us;liq.), lb/bbl (us;beer),lb/bbl (us;oil), lb/bbl (us;tank)

Pound/volume unit

Massoz, lb, STon

Ounce, pound, standard ton

Mass flow

oz/s, oz/min, oz/h, oz/dOunce/time unit

lb/s, lb/min, lb/h, lb/dPound/time unit

STon/s, STon/min, STon/h, STon/dStandard ton/time unit

Temperature°F, °R

Fahrenheit, Rankine

Volumeaf

Acre footft³

Cubic footfl oz (us), gal (us), kgal (us), Mgal
(us)

Fluid ounce, gallon, kilogallon, million gallonbbl (us;liq.), bbl (us;beer), bbl(us;oil), bbl (us;tank)Barrel (normal liquids), barrel (beer), barrel(petrochemicals), barrel (filling tanks)

af/s, af/min, af/h, af/dAcre foot/time unit

ft³/s, ft³/min, ft³/h, ft³/d

Cubic foot/time unit

fl oz/s (us), fl oz/min (us), fl oz/h(us), fl oz/d (us)Fluid ounce/time unit

gal/s (us), gal/min (us), gal/h (us),gal/d (us)Gallon/time unit

kgal/s (us), kgal/min (us), kgal/h(us), kgal/d (us)Kilogallon/time unit

Volume flow

90

Proline Promag 10 IO-Link

Endress+Hauser

Proline Promag 10 IO-Link

Explanation of abbreviated units

Process variable Units

Time

6.3

Mgal/s (us), Mgal/min (us),Mgal/h (us), Mgal/d (us)Million gallon/time unit

bbl/s (us;liq.), bbl/min (us;liq.),bbl/h (us;liq.), bbl/d (us;liq.)

Barrel/time unit (normal liquids)
Normal liquids: 31.5 gal/bbl

bbl/s (us;beer), bbl/min (us;beer),bbl/h (us;beer), bbl/d (us;beer)

Barrel /time unit (beer)
Beer: 31.0 gal/bbl

bbl/s (us;oil), bbl/min (us;oil),bbl/h (us;oil), bbl/d (us;oil)Barrel/time unit (petrochemicals)Petrochemicals: 42.0 gal/bbl

bbl/s (us;tank), bbl/min (us;tank),
bbl/h (us;tank), bbl/d (us;tank)Barrel/time unit (filling tank)
Filling tanks: 55.0 gal/bbl

s, m, h, d, ySecond, minute, hour, day, year

am, pmAnte meridiem ( before midday), post meridiem (after
midday)Imperial units

Process variable Units

Explanation

Densitylb/gal (imp), lb/bbl (imp;beer),lb/bbl (imp;oil)Pound/volume unit

Volumegal (imp), Mgal (imp)

Gallon, mega gallonbbl (imp;beer), bbl (imp;oil)Barrel (beer), barrel (petrochemicals)

gal/s (imp), gal/min (imp), gal/h(imp), gal/d (imp)

Gallon/time unit

Mgal/s (imp), Mgal/min (imp),
Mgal/h (imp), Mgal/d (imp)

Mega gallon/time unit

bbl/s (imp;beer), bbl/min(imp;beer), bbl/h (imp;beer), bbl/d(imp;beer)

Barrel /time unit (beer)
Beer: 36.0 gal/bbl

bbl/s (imp;oil), bbl/min (imp;oil),bbl/h (imp;oil), bbl/d (imp;oil)Barrel/time unit (petrochemicals)Petrochemicals: 34.97 gal/bbl

s, m, h, d, ySecond, minute, hour, day, year

am, pmAnte meridiem ( before midday), post meridiem (after
midday)Volume flow

Time

Endress+Hauser

Explanation

91

Index

Proline Promag 10 IO-Link

Index
A

Activate SW option (Parameter) . . . . . . . . . . . . . . . . 89
Active diagnostic IO-Link (Parameter) . . . . . . . . . . . . 25
Active diagnostics (Submenu) . . . . . . . . . . . . . . . . . . 25
Actual diagnostics (Parameter) . . . . . . . . . . . . . . . . . 25
Alarm delay (Parameter) . . . . . . . . . . . . . . . . . . . . . . 36
Application (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Application specific tag (Parameter) . . . . . . . . . . . . . . 70
Assign behavior of diagnostic no. 043 (Parameter) . . . 37
Assign behavior of diagnostic no. 230 (Parameter) . . . 38
Assign behavior of diagnostic no. 231 (Parameter) . . . 38
Assign behavior of diagnostic no. 302 (Parameter) . . . 38
Assign behavior of diagnostic no. 376 (Parameter) . . . 38
Assign behavior of diagnostic no. 377 (Parameter) . . . 39
Assign behavior of diagnostic no. 832 (Parameter) . . . 40
Assign behavior of diagnostic no. 833 (Parameter) . . . 41
Assign behavior of diagnostic no. 834 (Parameter) . . . 41
Assign behavior of diagnostic no. 835 (Parameter) . . . 42
Assign behavior of diagnostic no. 842 (Parameter) . . . 42
Assign behavior of diagnostic no. 937 (Parameter) . . . 43
Assign behavior of diagnostic no. 938 (Parameter) . . . 43
Assign behavior of diagnostic no. 961 (Parameter) . . . 44
Assign behavior of diagnostic no. 962 (Parameter) . . . 44
Assign process variable 1 to n (Parameter) . . . . . . 13, 54
Assign simulation process variable (Parameter) . . . . . 33

BBluetooth (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 77
Bluetooth configuration (Submenu) . . . . . . . . . . . . . . 77
Brightness (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 88

C

Calibration (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 65
Calibration factor (Parameter) . . . . . . . . . . . . . . . . . . 66
Clear event list (Parameter) . . . . . . . . . . . . . . . . . . . . 32
Color scheme (Parameter) . . . . . . . . . . . . . . . . . . . . . 88
Commissioning (Menu) . . . . . . . . . . . . . . . . . . . . . . . . 7
Communication established (Parameter) . . . . . . . . . . 77
Conductivity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 46
Conductivity calibration factor (Parameter) . . . . . . . . 66
Conductivity damping time (Parameter) . . . . . . . . . . . 60
Conductivity measurement (Parameter) . . . . . . . . . . . 59
Conductivity temperature coefficient (Parameter) . . . . 60
Conductivity unit (Parameter) . . . . . . . . . . . . . . . 12, 52
Configuration counter (Parameter) . . . . . . . . . . . . . . 73
Confirm Maintenance code (Parameter) . . . . . . . . . . . 76
Connectivity (Submenu) . . . . . . . . . . . . . . . . . . . . . . 77
Corrected conductivity (Parameter) . . . . . . . . . . . . . . 46

D

Date / Time (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 78
Date/time (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . . 22
Define Maintenance code (Parameter) . . . . . . . . . . . . 76
Define Maintenance code (Wizard) . . . . . . . . . . . . . . 76
Density unit (Parameter) . . . . . . . . . . . . . . . . . . . 11, 51
Device (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
92

Device alarm simulation (Parameter) . . . . . . . . . . . . . 34
Device ID (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 69
Device identification (Wizard) . . . . . . . . . . . . . . . . . . . 7
Device management (Submenu) . . . . . . . . . . . . . . . . . 72
Device name (Parameter) . . . . . . . . . . . . . . . . . . . . 8, 80
Device reset (Parameter) . . . . . . . . . . . . . . . . . . . . . . 73
Device tag (Parameter) . . . . . . . . . . . . . . . . . . . . 72, 81
Diagnostic 1 IO-Link (Parameter) . . . . . . . . . . . . . . . . 29
Diagnostic 2 IO-Link (Parameter) . . . . . . . . . . . . . . . . 29
Diagnostic 3 IO-Link (Parameter) . . . . . . . . . . . . . . . . 30
Diagnostic 4 IO-Link (Parameter) . . . . . . . . . . . . . . . . 31
Diagnostic 5 IO-Link (Parameter) . . . . . . . . . . . . . . . . 31
Diagnostic configuration (Submenu) . . . . . . . . . . . . . 36
Diagnostic event simulation (Parameter) . . . . . . . . . . 34
Diagnostic list (Submenu) . . . . . . . . . . . . . . . . . . . . . 28
Diagnostic settings (Submenu) . . . . . . . . . . . . . . . . . . 36
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Diagnostics 1 (Parameter) . . . . . . . . . . . . . . . . . . . . . 28
Diagnostics 2 (Parameter) . . . . . . . . . . . . . . . . . . . . . 29
Diagnostics 3 (Parameter) . . . . . . . . . . . . . . . . . . . . . 30
Diagnostics 4 (Parameter) . . . . . . . . . . . . . . . . . . . . . 30
Diagnostics 5 (Parameter) . . . . . . . . . . . . . . . . . . . . . 31
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Display (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19
Display damping (Parameter) . . . . . . . . . . . . . . . . 22, 88
Display module (Submenu) . . . . . . . . . . . . . . . . . . . . 83
Document
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

E

ECC duration (Parameter) . . . . . . . . . . . . . . . . . . . . . 67
ECC interval (Parameter) . . . . . . . . . . . . . . . . . . . . . . 67
ECC polarity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 68
ECC recovery time (Parameter) . . . . . . . . . . . . . . . . . 67
Electrode cleaning cycle (Parameter) . . . . . . . . . . . . . 67
Electrode cleaning cycle (Submenu) . . . . . . . . . . . . . . 66
Electronics (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 37
Empty pipe adjust value (Parameter) . . . . . . . . . . 19, 63
Empty pipe detection (Parameter) . . . . . . . . . . . . 19, 63
Empty pipe detection (Submenu) . . . . . . . . . . . . . . . . 62
ENP version (Parameter) . . . . . . . . . . . . . . . . . . . . . . 82
Enter access code (Parameter) . . . . . . . . . . . . . . . . . . 75
Event logbook (Submenu) . . . . . . . . . . . . . . . . . . . . . 32
Extended order code 1 (Parameter) . . . . . . . . . . . . . . 82
Extended order code 2 (Parameter) . . . . . . . . . . . . . . 82
Extended order code 3 (Parameter) . . . . . . . . . . . . . . 82

F

Filter options (Parameter) . . . . . . . . . . . . . . . . . . . . . 32
Firmware version (Parameter) . . . . . . . . . . . . . 8, 81, 83
Fixed density (Parameter) . . . . . . . . . . . . . . . . . . . . . 61
Flow damping (Parameter) . . . . . . . . . . . . . . . . . 16, 59
Flow override (Parameter) . . . . . . . . . . . . . . . . . . . . . 59
Full pipe adjust value (Parameter) . . . . . . . . . . . . 19, 64
Function tag (Parameter) . . . . . . . . . . . . . . . . . . . . . 70

Endress+Hauser

Proline Promag 10 IO-Link

Index

Guidance (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

GReset Maintenance code (Parameter) . . . . . . . . . . . . . 75
Rotation display (Parameter) . . . . . . . . . . . . . . . . . . . 88

H

S

Heartbeat Technology (Submenu) . . . . . . . . . . . . . . . 35

I

Information (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 80
Installation direction (Parameter) . . . . . . . . . . . . . . . 64
Integration time (Parameter) . . . . . . . . . . . . . . . . . . . 65
IO-Link (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 69

L

Language (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 85
Last diagnostic IO-Link (Parameter) . . . . . . . . . . . . . . 26
Location tag (Parameter) . . . . . . . . . . . . . . . . . . . . . . 70
Locking status (Parameter) . . . . . . . . . . . . . . . . . . . . 72
Low flow cutoff (Parameter) . . . . . . . . . . . . . . . . 16, 61
Low flow cutoff (Submenu) . . . . . . . . . . . . . . . . . . . . 61

M

Manufacturer (Parameter) . . . . . . . . . . . . . . . . . . . . . 83
Mass flow (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 45
Mass flow unit (Parameter) . . . . . . . . . . . . . . . . . 11, 51
Measured value EPD (Parameter) . . . . . . . . . . . . . . . . 64
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 45
Measuring conditions (Wizard) . . . . . . . . . . . . . . . . . 15
Measuring period (Parameter) . . . . . . . . . . . . . . . . . . 65
Menu
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Commissioning . . . . . . . . . . . . . . . . . . . . . . . . . . . 7
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Guidance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 71

N

New adjustment (Parameter) . . . . . . . . . . . . . . . . . . . 63
Nominal diameter (Parameter) . . . . . . . . . . . . . . . . . 65

O

Off value low flow cutoff (Parameter) . . . . . . . . . . 17, 62
On value low flow cutoff (Parameter) . . . . . . . . . . 17, 62
Operating time (Parameter) . . . . . . . . . . . . . . . . . . . . 27
Operating time from restart (Parameter) . . . . . . . . . . 26
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 81

P

Preset value 1 to n (Parameter) . . . . . . . . . . . . . . . . . 56
Pressure shock suppression (Parameter) . . . . . . . . . . 18
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . . 26
Process (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Process parameters (Submenu) . . . . . . . . . . . . . . . . . 58
Process value (Parameter) . . . . . . . . . . . . . . . . . . . . . 33
Process variable unit 1 to n (Parameter) . . . . . . . . 13, 54
Product ID (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 70
Product name (Parameter) . . . . . . . . . . . . . . . . . . . . 69
Properties (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 36

R

Reset all totalizers (Parameter) . . . . . . . . . . . . . . . . . 53

Endress+Hauser

Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 37, 58
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . . . . 64
Sensor electronic module (ISEM) (Submenu) . . . . . . . 83
Serial number (Parameter) . . . . . . . . . . . . . . . . . . . 8, 81
Set date/time (Parameter) . . . . . . . . . . . . . . . . . . . . . 23
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 33
Software configuration (Submenu) . . . . . . . . . . . . . . . 89
Software option overview (Parameter) . . . . . . . . . . . . 89
Submenu
Active diagnostics . . . . . . . . . . . . . . . . . . . . . . . . 25
Bluetooth configuration . . . . . . . . . . . . . . . . . . . . 77
Calibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 65
Connectivity . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 77
Date / Time . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 78
Device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
Device management . . . . . . . . . . . . . . . . . . . . . . . 72
Diagnostic configuration . . . . . . . . . . . . . . . . . . . . 36
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
Diagnostic settings . . . . . . . . . . . . . . . . . . . . . . . . 36
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Display module . . . . . . . . . . . . . . . . . . . . . . . . . . . 83
Electrode cleaning cycle . . . . . . . . . . . . . . . . . . . . 66
Electronics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Empty pipe detection . . . . . . . . . . . . . . . . . . . . . . 62
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Heartbeat Technology . . . . . . . . . . . . . . . . . . . . . 35
Information . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
IO-Link . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 69
Low flow cutoff . . . . . . . . . . . . . . . . . . . . . . . . . . 61
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 45
Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Process parameters . . . . . . . . . . . . . . . . . . . . . . . 58
Properties . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 37, 58
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 64
Sensor electronic module (ISEM) . . . . . . . . . . . . . . 83
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33
Software configuration . . . . . . . . . . . . . . . . . . . . . 89
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Totalizer handling . . . . . . . . . . . . . . . . . . . . . . . . 53
Totalizers . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
User management . . . . . . . . . . . . . . . . . . . . . . . . 74
System (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 71
System units (Submenu) . . . . . . . . . . . . . . . . . . . . . . 49
System units (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . 8

TTarget group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Temperature (Parameter) . . . . . . . . . . . . . . . . . . . . . 47
Temperature damping time (Parameter) . . . . . . . . . . 60
Temperature unit (Parameter) . . . . . . . . . . . . . . . 12, 52
Time format (Parameter) . . . . . . . . . . . . . . . . . . . . . . 22
Time zone (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 22
93

Index

Proline Promag 10 IO-Link

Timestamp (Parameter) . . . . . . . . . . . . . . 26, 29, 30, 31
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 47
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . . . . 53
Totalizer 1 to n (Wizard) . . . . . . . . . . . . . . . . . . . . . . 13
Totalizer 1 to n control (Parameter) . . . . . . . . . . . . . . 56
Totalizer 1 to n failure behavior (Parameter) . . . . 15, 57
Totalizer 1 to n operation mode (Parameter) . . . . 14, 55
Totalizer 1 to n overflow (Parameter) . . . . . . . . . . . . . 48
Totalizer 1 to n value (Parameter) . . . . . . . . . . . . . . . 47
Totalizer handling (Submenu) . . . . . . . . . . . . . . . . . . 53
Totalizers (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 53

U

User management (Submenu) . . . . . . . . . . . . . . . . . . 74
User role (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 74

V

Value 1 display (Parameter) . . . . . . . . . . . . . . . . . 20, 86
Value 2 display (Parameter) . . . . . . . . . . . . . . . . . 20, 86
Value 3 display (Parameter) . . . . . . . . . . . . . . . . . 21, 87
Value 4 display (Parameter) . . . . . . . . . . . . . . . . . 21, 87
Vendor name (Parameter) . . . . . . . . . . . . . . . . . . . . . 69
Volume flow (Parameter) . . . . . . . . . . . . . . . . . . . . . 46
Volume flow unit (Parameter) . . . . . . . . . . . . . . . . 9, 49

W

WizardDate/time . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22
Define Maintenance code . . . . . . . . . . . . . . . . . . . 76
Device identification . . . . . . . . . . . . . . . . . . . . . . . . 7
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19
Measuring conditions . . . . . . . . . . . . . . . . . . . . . . 15
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . . 13

Z

Zero point (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 66

94

Endress+Hauser

*71654754*
71654754

www.addresses.endress.com