Handheld Particle Counter
MODEL 3888/3889

User ManualiList of Components■ Standard
Items
Body

MODEL
Body (or 3889)

3888

AC Adaptor
USB Cable

3888-103888-20

Zero Count Filter with tubing
Inlet Protective Cap

Functions3888-60
3888-61

18650 Lithium ion battery

Operates the instrument with AC power andcharge the internal rechargeable battery
Communicates with PC or PrinterCleans the air flow path inside the instrumentwith clean air
Keeps out dust and contaminants from the
instrument when not in use
Supplies power to the instrument
Nominal voltage:3.7V-

Rated capacity:2600mAh min.
Outline size:18mm
Length:69mm
With a protection function

Quick Start Guide
Measurement Software

3888-40■ Optional Accessories
Items
Temperature/Humidity
Probe
CradleIsokinetic Suction Nozzle

MODEL
0842
3888-70
3887-04

Functions
Measures temperature and humidity
Stands the instrument and perform
Ethernet/Wi-Fi/RS485 communications
To be connected to the inlet to match themeasurement condition with the sampling air

Carrying Case

3888-71

TripodEX-344Q

Printer

DPU-S24500B

Prints the measured data directly from the

TP-202L

A 10-roll package

Printer Roll Paper

Stores the instrument

instrumentiImportant Safety Information
The symbols for the warnings used in this manual are defined below:
Classifications
Warning
Warnings in this classification indicate risks that may result in serious injury
or death if not observed.
CautionWarnings in this classification indicate risks that may result in injury or
damage to the surrounding objects if not observed.
NoticeWarnings in this classification indicate risks of damage to the product that
may void the product warranty if not observed.

Description of Symbols
This symbol indicates a condition that requires caution (including warning).
The subject of each caution is illustrated inside the triangle (e.g. the hightemperature caution symbol is shown on the left).
This symbol indicates a prohibition. Do not take the prohibited action shown
inside or near this symbol (e.g. the disassembly prohibition symbol is shownon the left).
This symbol indicates a mandatory action. A specific action is given near thesymbol.
This symbol indicates a warning of possible laser radiation.

Warning（Forbidden）Do not use the AC adaptor other than the provided one with the
instrument.
Using an inappropriate adaptor may damage the instrument.
It may generate heat and cause fire.

（Do not remodel/disassemble）Never disassemble, modify, or repair.
This instrument uses a Class 3B laser diode as the light source.
Exposure to the laser may cause loss of eyesight and other injury.
Never open the instrument other than the battery compartment.
（Proper Handling）Handle the instrument properly in accordance with theiiinstructions provided in this manual.
Failure to do so may cause electric shock, fire, or sensor damage.

（Caution）If abnormal noise, smell, or smoke is observed, or if liquid hasentered the instrument, turn off the power immediately, remove the batteriesor pull out the plug.
Failure to observe the above may cause electric shock, fire hazard, or
damage.
Please contact your local distributor or Kanomax service center for repair.

Caution（Handle properly）Pull out the plug when the instrument is not in use.
Failure to do so may cause electric shock, fire hazard, and circuit damage.

(Proper Handling) Handle the batteries properly in accordance with thefollowing instructions.
1. Only use 18650 Lithium ion batteries. Do not use batteries other than
18650 Lithium ion batteries.
2. Be sure to insert the positive and the negative ends of the batteries in the3.

correct direction. Inserting batteries in the wrong direction may cause
battery leakage, overheating, explosion and/or fire.

Do not place the product and/or the batteries in high temperatureenvironments, or near any heat source. This may cause battery leakage,overheating, explosion and/or fire.

Notice（Forbidden）Do not use or keep the instrument in hot, humid, or dusty
environment. Do not expose the instrument to direct sunlight for a
prolonged period of time.
The instrument may not function properly out of the specified operationaltemperature range.
（Forbidden）Do not subject the instrument to strong impact.
Dropping or hitting the instrument may cause damage and malfunction.
（Forbidden）Do not wipe the instrument with volatile solvent.
The body may deform or deteriorate. Use soft dry cloth to remove stains.
If stains persist, soak the cloth in neutral detergent and wipe the instrumentwith the cloth.

iii

（Forbidden）Do not touch the LCD screen with a sharp-pointed object orwith excessive pressure.
It may cause distortion of the screen or a malfunction.
A rapid temperature change may cause a malfunction of the screen.
（Proper Handling）When storing the instrument, put the instrument in the
carrying bag and keep it in a place with an ambient temperature of -10 to
50℃ and no condensation.
（Forbidden）Do not dispose of the instrument as household waste.
Please note that the disposal of the instrument and batteries should be in linewith your local or national legislations. For details, please consult with yourlocal distributor.
(Proper Handling) Be sure to use the provided AC adaptor properly.
1. Prior to use, confirm that the voltage of the power supply outlet is within
the range of the power supply voltage specifications.
2. Use only grounded electrical outlets.
3. When the product is not used for a long period of time, or when the
product has to be powered off in urgent cases, remove the AC adaptor
from the outlet.
4. Do not use the AC adaptor in high temperatures and high humidity. Only
use the AC adaptor under indoor environments at an attitude of 2,000
meters or less.
5. If the Power cable or the AC adaptor is damaged, contact to your local
distributor or KANOMAX service center for repair.

iv

Table of Contents
Part Names and Functions ................................................................................................6
Getting Started ...................................................................................................................7
Charging Batteries .................................................................................................. 7
Caution ................................................................................................................. 7
Checking the Temperature/Humidity Probe（Optional accessory for the MODEL 3889） .. 7
Turning the Power ON/OFF ...................................................................................... 7

Measurement Procedures .................................................................................................7
Turning the Power ON ............................................................................................. 7
Internal Cleaning .................................................................................................... 7
Measurement ......................................................................................................... 8
Turning the Power OFF ............................................................................................ 8
Errors.................................................................................................................... 8

User Interface .....................................................................................................................9
Main Screen ........................................................................................................... 9
General setting ..................................................................................................... 11
Measurement setting ............................................................................................ 13
Preset ................................................................................................................. 15
History ................................................................................................................ 15
Print Example....................................................................................................... 16
Screen Transition .................................................................................................. 18

Specifications ...................................................................................................................19
Troubleshooting ...............................................................................................................21
Warranty and After Service .............................................................................................22
Contact Information .........................................................................................................23v

Part Names and Functions

The cradle is an optional accessary.
Names
1. Inlet

Functions
2.83L/min suction volume
Outer diameter 6.4mm

2. Power/Home button

Turns the power ON/OFF
To return to the Main screen in operation

3. USB port (for PC)

Connects with your PC by using the USB
cable

4. USB port (for Printer, for USB flash
memory)Connects with the printer by using theUSB cable
Allows you to copy the measurement
results to the USB flash drive

5. DC jack

Supplies power with the provided AC
adaptor6. Threaded tripod mountAttaches instrument to the tripod

7. Battery compartment

The battery is replaceable

8. Temperature/Humidity probe terminal

Connects the temperature/humidity(only for 3889)
9. Communication port with cradle

Communicates with the cradle

10. Communication port

RS-485 port

11. Communication port

Communicates with the Ethernet

12. Communication port with the Main

Communicates with the main body

body

6

probe

Getting Started
Charging BatteriesThis instrument operates on the internal rechargeable battery or the provided AC adaptor.
Connect the provided AC adaptor to start charging. It takes approximately 5 hours tocharge the battery completely.



If the batteryʼs run-time becomes shorter, the battery is replaceable.

Open thecompartment to replace the batteries with new 18650 Lithium-ion rechargeable batteries
with protection circuit. (69mm length).
Caution

Rapid temperature changes may cause measurement errors.

When moving the

instrument from one location to another with different ambient temperature allow a
sufficient time for the instrument to return to room temperature. (more than ten
minutes )Checking the Temperature/Humidity Probe（Optional accessory for the MODEL 3889）


Prolonged measurement under high temperature conditions or measurement under rapid
temperature change may result in abnormally high humidity readings due to condensation.
If condensation occurs, leave the probe in atmosphere of 40%RH or less for 24 hours to
dry the probe.

Turning the Power ON/OFF

Press and hold the Home button to turn the power on. A logo appears and then it turns
to the startup screen.

When the pump and other parts are ready to perform ameasurement, the Main screen for ready to measure will be displayed.


To turn the power OFF, press and hold the Home button (3 second or more) from any
screens.

Measurement Procedures
Turning the Power ONPress and hold the Home button to turn the power on. A logo appears and then it turns
to the startup screen. Touch the startup screen to display the Main screen ready tomeasure.

Internal Cleaning

Prior to use, purge the instrument (for internal cleaning) with the provided zero filter.

Remove the Inlet protective cap, and attach the provided zero count filter to the inlet.

Turn the power on and press the Start button. Perform a continuous measurement for
10 to 15 minutes until the measuring count value is suitable for 10 sec. or more.

After measurement, remove the zero count filter from the inlet.

7

Measurement
Use the General setup and Measurement setup sub-menus to configure the necessary
settings.



Set the measurement conditions from the Measurement mode. You can also select themeasurement conditions from the Preset.

Return to the Main screen.

Press the Start button to perform a measurement.

The measurement will complete automatically or by pressing the Stop button.



The previous measurement results can be viewed from the History sub-menu.

Turning the Power OFFPress and hold the Home button from any screens to turn the power OFF.

Errors

The following table explains possible errors due to self-diagnosis function. Even if an error
occurs, the measurement and other processes will be continued.
Laser power
failureFailure or end-of-life of the laser diodeThe pump current has increased or decreased.

Flow rate

Remove the inlet cap and filter from the instrument.
If the error persists, the possible cause is failure or theend-of-life of the pump.

Maximum
measurable
concentration
exceeded8

The concentration exceeds the maximum measurable
range. Perform the measurement at a cleaner location
or with the Zero count filter

User Interface
Main ScreenThere are four configurations for the main screen that can be selected to best suit yourpurposes. (The figures below show the user interfaces of 6 channel Model 3889.) Fromeach Main screen, you can move to sub-menu screens.

Table
Cumulative

Bar

Table
Differential

Circle9

Icons

Names
General setting

Functions
Opens the menus to set the unit, calendar, data,sound,

languageandtodisplay

instrument

information
Measurement

Opens the menu to set the mode, alarm, save,setting

remote and temperature/humidity

PresetOpens the menu to set or select up to 18 Presets

History

Views the historical data or measurements

Change chart

Switches the chart types:

types

Bar → Circle → Table ∑∑∑Δ Display

Δ∑Δ Display

Switching

Switching

Switches between Cumulative ∑ and Differential Δ

Switches between Cumulative ∑ and Differential Δ

Start

Starts a measurement

Stop

Stops a measurement

Particle Size
Setup

Particle Size
Setup

Error
Battery Level

Tap the chart window to set the required particle size

Tap the circle chart to set the required particle size

Tap the Error to display an error description
Icons show how much battery is remaining. AC adapter
⇒

10

60%⇒

40%⇒

10%

operatesIcons

Names

Functions
Displays the selected measurement modes
Display

Mode

Mode

Main

RepeatRepeatSingleSingleContinuous

Continuous

Calculation

Calculation

ISO

ISO

GB

GB

Display
Displays the selected preset modes
Display
Mode

Preset 1

Preset 2

Preset 3

RepeatP1(Repeat)

P2(Repeat)

P3(Repeat)

SingleP1(Single)

P2(Single)

P3(Single)

Continuous

P1(Continuous) P2(Continuous) P3(Continuous)

Calculation

P1(Calculation)

P2(Calculation)

P3(Calculation)ISO

P1(ISO)P2(ISO)P3(ISO)GB

P1(GB)

P2(GB)

P3(GB)

General setting

Select the [General setting] icon on the Main screen to set the unit, calendar, data,
language, and sound. This menu indicates the instrument information.

Icons

Names

Functions
Select one of the following 3 options of units to reflect to thereadings displayed on the Main screen and Measurement

Unit

results.


count︓Measured particle count/m3: Particle count per cubic meter/ft3: Particle count per cubic footSet the date and time
Calendar

Touch the setup button to display the numeric keypad.
Enter 2 digits in blanks for year, month, day, hour, andminute.

11

Icons

Names

Functions
Confirms the information and perform operation on the entire
measured data

Memory remaining: Displays the ratio of the free space to thewhole memory capacity.

Number of Records: Displays the number of the stored
records.



Copy: Copies all files to the USB flash memory.

Delete: Deletes all files.

It is therefore recommended tocopy the files prior to use this function.
Note that copying the measured data to the USB flash memoryand deleting from the internal memory are not partial but batchprocessing. Follow the following procedure.
Data

Step 1

Step 2

Step 3

Select [Data] 【Copy the measured data toin thethe USB Flash Memory】

GeneralTap [YES] tosetting

execute.
Insert the USB flash memory
into the main body and tap
the [Copy all] button.
【Delete the measured Data】

Tap [NO] toreturn without
running thecommand.

Tap the [Delete all] button.

LanguageSound

Selects a language
English, Japanese, Chinese, Spanish

Sets the operation sound ON/OFF
Displays the instrument information

Information

Latest calibration date: Depending on the usage and condition, it
is normally recommended to calibrate at least once a year.
Version: To show firmware version of the instrument.

12

Measurement setting

Select [Measurement Setup] in the Main screen to set the mode, alarm, save, remote, andtemperature/humidity.

Icons

Names

Functions
Selects 6 options of the measurement modes. Measurements will be
performed under the latest mode setting.


Repeat mode: Repeats the measurement of configured samplingtime and cycle.

Single mode: Performs a measurement of set sampling time only
once.



Continuous mode: Continues a measurement

until pressing theStop button

Calculation mode: Repeats the measurement under the same
condition of the Repeat mode and then calculate the average,
standard deviation, maximum, and minimumISO mode: Performs measurements in accordance with ISO 14644



GB mode: Performs measurements in accordance with Chinese
National Standard
Set the required conditions for 6 options of measurement modes.
location

Mode

Sampling

Repeat countInterval time

time

Repeat✔

✔

✔

✔

Single✔

✔

――Continuous

✔

―

―

―

Calculation

✔

✔

✔

✔

ISO

✔

✔

✔

✔

GB

✔

✔

✔

✔

Tap indication () to display a Ten keyboard.

Set the measurement conditions by entering numbers.

13

ItemsDescription

Range

Location

Sets measurement locations by number.

From 1 to 99

Sampling Sets the sampling time of 1 cycle.

From 6 sec. totime

99 min. 59
sec.

Repeat

Sets the number of repeat.

From 2 to 999

countInterval
time

Sets the interval between the starting time From 6 sec. toof a measurement and the starting time of 99 min. 59
the next measurement.

sec.

The repeat interval must be longer thansampling time.

Difference between

interval time and sampling time issuspension time.

Icons

Names

Functions
Configures the alarm setting for the measured values in the
Cumulative value. Tap indication () for each particle size

to configure the following settings.

Alarm

Alarm: To select the alarm function ON/OFF (Default
setting: OFF）



Threshold: To set the threshold value for the measured
cumulative value for each particle size (Default: No
setting）
The threshold value you set here will be the upper limit of
the chart display.

Save

Toggles between saving or not saving the measurements

Use this setting to perform measurements according tocommands from a remote computer
Remote

Setting: sets the communication method for the "RS485","Ethernet", "Wi-Fi" and each connection methodʼsparameters. ID is the identification number of the device.


Temperature/
Humidity

14

Connection: starts to connect outer PC

Sets measurement with the temperature/humidity probe
ON/OFF and set the temperature unit (°C or °F)(Only for 6-channel model 3889）

Preset

Select [Preset] in the Main screen to set and select measurement conditions.



Three options of Preset number provide 6 measurement modes each; therefore 18
measurement conditions in total can be preset.

When setting the Preset, select one of the 18 measurement modes in the Step 3, then set
the measurement conditions in the Step 4 as shown below:
Step 1

Step 2

Step 3

Step 4

SelectSelect one Preset number.

Select the Mode.

Set or confirm the[Preset] inNo measurement

Measurement

measurement

the Mainconditions have been

conditions have been

conditions. Tap ✔

Screen.

selected yet.

selected here.

select mode.

toHistory

Select [History] in the Main screen to confirm the previous measurement results.

Step 1

Step 2

SelectSelectthe

The measurement[History] in themeasurement modes.

results is displayed.

Main screen.

one

Step 3ofDisplays the historyof the previous number.
Displays the historyof the next number.
Selects anddisplay the history of
the arbitral number.
Selects the unit.
Prints the result.

15Print Example
The following figures show print examples of 6 channel model.

(1) Repeat mode(3) Calculation mode

2017/3/30 11: 02
RepeatE= - - Number

1

2017/3/30 11: 02
Calculation

7

Location1

Location

1

S-Time00:21

S-Time00:21

I -Time

00:21
87CNT

I -Time

00:21

0.3um

16

CNT

SD

128CNT

0.5um

14

CNT

MAX

235CNT

1.0um

12

CNT

3.0um

2

CNT

5.0um

1

CNT

(2) Single, Continuous mode

Single

0.3um

0.5um

1.0um

2017/3/30 11: 02

E= - - Number

1

Location

1

S-Time
0.3um

AVE

MIN

0CNT

AVE

39CNT

SD

66CNT

MAX

116CNT

MIN

0CNT

AVE

12CNT

SD

19CNT

MAX

35CNT

MIN

0CNT

AVE

0CNT

00:21

SD

0CNT

16

CNT

MAX

1CNT

0.5um

14

CNT

1.0um

12

CNT

3.0um

2

5.0um
10.0um

3.0um

MIN

0CNT

AVE

0CNT

CNT

SD

0CNT

1

CNT

MAX

0CNT

1

CNT

5.0um

10.0um16

E= - - Number

MIN

0CNT

AVE

0CNT

SD

0CNT

MAX

0CNT

MIN

0CNT

(4) ISO mode(5) GB mode

2017/3/30 11: 02
ISO

E= - - Number

From

1

To

6

S-Time00:21

I -Time00:21-------------------------------Piont=001

Times=001

0.5um

43

/m3

5.0um

0

/m3

Piont=001

Times=002

0.5um

7

/m3

5.0um

0

/m3

Piont=002

Times=001

0.5um

7

/m3

5.0um

0

/m3

Piont=002

Times=002

0.5um

1

/m3

5.0um

0

/m3

Piont=003

Times=001

0.5um

5

/m3

5.0um

0

/m3

Piont=003

Times=002

0.5um

6

/m3

5.0um

0

/m3

-------------------------------0.5um

AVE

11

/m3

SD

6

/m3

2017/3/30 11: 02
GB

E= - - Number

From

1

To

6

S-Time00:21

I -Time00:21-------------------------------Piont=001

Times=001

0.5um
5.0um
Piont=001

43

/m3

0

/m3

Times=002

0.5um

7

/m3

5.0um

0

/m3

Piont=002

Times=001

0.5um
5.0um
Piont=002

7

/m3

0

/m3

Times=002

0.5um

1

/m3

5.0um

0

/m3

Piont=003

Times=001

0.5um
5.0um
Piont=003

5

/m3

0

/m3

Times=002

0.5um

6

/m3

5.0um

0

/m3

-------------------------------0.5um

AVE

11

/m3

SD

6

/m3

5.0um

AVE

0

/m3

SD

0

/m3

17

Screen Transition

The screens are structured by 6 layers and the operations are mainly performed from theLayer 2 and higher.



The Main screen for ready to measure is displayed when the instrument is not in themeasuring process.

The Main screen for measurement is displayed when performing

measurements.


From the Main screen before measurement, you can move to the 4 menu screens: General
setting, Measurement setting, Preset, and History.



Pressing the Home button on the main body will jump from a screen of any layers to theTable∑ on the Main screen.
Layer 1

Layer 2

Layer 3Layer 4Layer 5Layer 6Logo
Main (Table∑)

Measurement
ResultMain (TableΔ)

ERROR

StartupSound
Unit
CalendarMain (Bar)

Menu(GeneralSetup)

Information

Copy

Data

DeleteMain (Circle)
Language

Measurement
Mode

Continue/Stop

Menu(Measurement Setup)

RepeatAlarm

SingleSave

Continuous

RemoteCalculation

Tempreture/
Humidity

ISO
GB
Setup

Menu
(Preset)Menu(History)

18

1

Setup

2

Setup

3

Setup

Details

Confirm

Specifications
Product

Handheld Particle Counter
3888 (3-channel model)Model3889 (6-channel model)
Measuring
methodSize distribution

Light scattering
6 channels (0.3, 0.5, 1.0, 3.0, 5.0, 10.0µm）
3 channels (0.3, 0.5, 5.0µm）
0.1CFM (2.83L/min)

Flow rate

Accuracy: ±5% (Compliant with JIS B9921 andISO21501-4）

Sampling time

From 6 seconds to 99 minutes and 59 seconds（for 1 measurement cycle）
From 6 seconds to 99 minutes 59 seconds

Interval time

（Interval between the start and the next start of the repeat
measurements）

Sampling cycle
Locationclassification
Calibration
Particle
Measurement

From 1 to 999 cycles or continuous measurements
99 locations
NIST traceable

Measuring mode Repeat, Single, Continuous, Calculation, ISO, GB
Display time of
measured value
Display ofmeasured value
Maximum
measurable
Concentration

From 1 to 10 seconds
Differential Δ and Cumulative∑
2,000,000 particles/CF at 10% coincidence loss
（compliant with JIS B9921 and ISO21501-4）
50±20%（for PSL particles near the minimum measurable

Counting
efficiency

size）
100±10%（for PSL particles of 1.5 to 2 times as large as theminimum measurable size）（Compliant with JIS B9921 and ISO21501-4）

False countSize resolution

Display≦1 particle/5 minutes（Compliant with JIS B9921 and ISO21501-4）≦15% (for PSL particles near 0.3µm)（Compliant with JIS B9921 and ISO21501-4）

Pump

Internal pump（Vane）

ExhaustWith filter
4.3 inch color LCD, Resistive touch panel

19

Communication

RecordingmediaStandard

Device: for PC）

Cradle (Option)

Ethernet, Wi-Fi, and RS485 (9600, 19200, 38400 baud)

Media

Internal memory

Number andformatLanguagePower

USB (Host: for printer and USB flash memory,

Up to 10,000 records in CSV format
English, Japanese, Chinese, and Spanish

Internal supply

Li-ion rechargeable battery (Replaceable）

External supply

AC adaptor Input 100 to 240 V

Continuous
operation time

Up to 5 hours（Single measurement）

OperatingMain body

From 10 to 40 ℃, from 0 to 85%RH (With no condensation）

environment

Probe

From 0 to 50 ℃, from 2 to 98%RH（With no condensation）

Dimension

W100 X H213 X D55 mm （except Inlet）

Weight650g
Probe

Model 0842

Measurement（Temperature）From 0 to 50℃

range（Humidity）From 2.0 to 98.0％RH

Hydrothermal

Display（Temperature）0.1℃

measurement

resolution（Humidity）0.1％(Option)

Humidity(6 channel)

accuracy
Temperatureaccuracy
Response time

±3.0% (30 to 85%RH), ±5% (other range of humidity)±0.5℃
Approximately 60 seconds or less（90％ response）

Wi-Fi is a trademark or a registered trademark of Wi-Fi Alliance.

20Troubleshooting
Symptoms
The display does notappear when the power isturned ON.
The battery drains fast.

Particle count orconcentration is too high.

Particle count orconcentration is too low.
The printer does notoperate.

Possible causes(s) → Solution(s)The AC adapter is not connected properly.
→ Confirm the AC adapter and power cable.
Low battery→ Replace the batteries.
→ Recharge the batteries.
The battery is deteriorated.
→ Replace the battery
The actual concentration is high or the instrument may
malfunction.
→ Attach the provided Zero count filter and confirm
that the reading drops to zero. If the reading
remains high, the instrument may malfunction.
Please contact Kanomax USA.
Laser power failure or flow error→ The instrument may malfunction.
Please contact Kanomax USA.
Incorrect baud rate setting→ Confirm the printer setting.

Reference
6

6

7

-

Saving data function is OFF →Confirm the save setting.
The number of recorded data may exceed 10,000.

1210

If Wi-Fi remote connection is used above a metal
The measured value of the conductor, the Wi-Fi radio wave may be reflected and affectthe measured result.
particles maybe higher in→ When using Wi-Fi connection above any metal
Wi-Fi remote mode.
conductor, please use a non - metallic spacer of 3 cm or
more.

-

Data can not be stored.

21

Warranty and After Service
The limited warranty set below is given by KANOMAX USA, Inc. (hereafter referred to as “KUI”) withrespect to this instrument, its attachment parts including standard accessories (hereafter referred to as“PRODUCT”) that you have purchased. PRODUCT you have purchased shall be the only one that thelimited warranty stated herein applies to.
Your PRODUCT, when delivered to you in new condition in its original container, is warranted against
defects in materials or workmanship as follows:

for a period of two (2) years from the date of original

purchase, defective parts or a defective PRODUCT returned to KUI, as applicable, and proven to be
defective upon inspection, will be exchanged for a new or comparable rebuilt parts, or a refurbished
PRODUCT as determined by KUI.

Warranty for such replacements shall not extend the original warranty

period of the defective PRODUCT.
To obtain service under this warranty, you must notify Kanomax USA, Inc. on or before the expiration of
the warranty period to obtain directions for returning the defective product.

You are responsible for all

return shipping charges to the authorized Kanomax service center.
This limited warranty covers all defects encountered in normal use of the PRODUCT, and does not apply tothe following cases:(1) Use of parts or supplies other than the PRODUCT sold by KUI, which cause damage to the PRODUCTor cause abnormally frequent service calls or service problems.
(2) If any PRODUCT has its serial number or date altered or removed.
(3) Loss or damage to the PRODUCT due to abuse, mishandling, improper packaging by the owner,
alteration, accident, electrical current fluctuations, failure to follow operating, maintenance orenvironmental instructions prescribed in the PRODUCT's instruction manual provided by KUI, or service
performed by other than KUI.
NO IMPLIED WARRANTY, INCLUDING ANY IMPLIED WARRANTY OF MERCHANTABILITY OR FITNESS FOR
A PARTICULAR PURPOSE, APPLIES TO THE PRODUCT AFTER THE APPLICABLE PERIOD OF THE EXPRESSLIMITED WARRANTY STATED ABOVE, AND NO OTHER EXPRESS WARRANTY OR GUARANTY, EXCEPT AS
MENTIONED ABOVE, GIVEN BY ANY PERSON OR ENTITY WITH RESPECT TO THE PRODUCT SHALL BINDKUI.

KUI SHALL NOT BE LIABLE FOR LOSS OF STORAGE CHARGES, LOSS OR CORRUPTION OF DATA,

OR ANY OTHER SPECIAL, INCIDENTAL OR CONSEQUENTIAL DAMAGES CAUSED BY THE USE OR MISUSE
OF, OR INABILITY TO USE, THE PRODUCT, REGARDLESS OF THE LEGAL THEORY ON WHICH THE CLAIM
IS BASED, AND EVEN IF KUI HAS BEEN ADVISED OF THE POSSIBILITY OF SUCH DAMAGES.

IN NO

EVENT SHALL RECOVERY OF ANY KIND AGAINST KUI BE GREATER IN AMOUNT THAN THE PURCHASE
PRICE OF THE PRODUCT SOLD BY KUI AND CAUSING THE ALLEGED DAMAGE.

WITHOUT LIMITING

THE FOREGOING, THE OWNER ASSUMES ALL RISK AND LIABILITY FOR LOSS, DAMAGE OF, OR INJURY
TO THE OWNER AND THE OWNER'S PROPERTY AND TO OTHERS AND THEIR PROPERTY ARISING OUT OF
USE OR MISUSE OF, OR INABILITY TO USE, THE PRODUCT NOT CAUSED DIRECTLY BY THE NEGLIGENCEOF KUI.

THIS LIMITED WARRANTY SHALL NOT EXTEND TO ANYONE OTHER THAN THE ORIGINAL

PURCHASER OF THE PRODUCT, OR THE PERSON FOR WHOM IT WAS PURCHASED AS A GIFT, AND
STATES THE PURCHASER'S EXCLUSIVE REMEDY.

22

Contact Information



U.S.A.
KANOMAX USA, INC.
219 US Hwy 206, Andover, New Jersey 07821 U.S.A.
TEL: (800)-247-8887 / (973)-786-6386
FAX: (973)-786-7586
URL: http://www.kanomax-usa.com/
E-Mail: info@kanomax-usa.com



JAPAN
KANOMAX JAPAN INC.
2-1 Shimizu, Suita City, Osaka 565-0805, Japan
TEL: 81-6-6877-0183
FAX: 81-6-6879-5570
URL: http://www.kanomax.co.jp
E-Mail: sales@kanomax.co.jp



CHINA
Shenyang Kano Scientific Instrument Co., Ltd.
TEL: 86-24-23846440
FAX: 86-24-23898417
URL: http://www.kanomax.com.cn/
E-mail: sales@kanomax.com.cn

06001
19.01

23

24