EN

INSTRUCTIONSDATA AND REFERENCESQUESTIONS AND ANSWERS
IMPORTANT:READ ALL INSTRUCTIONS THOROUGHLY BEFORE USE.
KEEP THIS LEAFLET AND REFER TO IT PERIODICALLY.

Related Products of Super-Bond(sold separately)

Dental Adhesive Resin Cement

Porcelain Liner M

FOR DENTIST USE ONLY

Super-Bond

Adhesive primer for porcelain

V-PRIMER

Adhesive primer for precious metal alloys

Super-Bond SEP

Water Soluble release agent

Accessories for Super-Bond(sold separately)
Dispensing Dish (Ceramic)
Dispensing Stand
Dispensing Cups 40 pieces
Sponge (L • S)
Measuring Spoon (Standard) for the standard mix ratio
Measuring Spoon (Small) for 75% of the standard mix ratio
Measuring Spoon (Large) for 120% of the standard mix ratio
Brush Handle (Straight)
Brush Handle (Bent)
Brush Tips (Blue • Bulk-mix)
Brush Tips (Green • S • Brush-dip)
Brush Tips (Pink • L • Brush-dip)
Brush Tips (Purple • LL • Brush-dip)
Needle Tips (23G) 50 pieces and Needle Caps (Red) 2 pieces
Needle Tips (23G) 50 pieces and Needle Caps (Green) 2 pieces

Date of issue: 2020-08-06
J026E/1CTable of Contents
Instructions

CAUTION
1.
What is Super-Bond?
2.
Contents of Super-Bond kits
3.
Precautions
3–1. Safety
3–2. Precautions on Catalyst V
3–3. Storage
3–4. To Get the Best Results with Super-Bond
4.
How to use Super-Bond
Surface Preparation
BULK-MIX TECHNIQUE
BRUSH-DIP TECHNIQUE
5.
Examples of Clinical Applications

Data and References

Table 1
Table 2
Table 3Fig. 1
Fig. 2
Table 4
Table 5
Table 6
Table 7
Table 8
Table 9Fig. 3
Table 10
Table 11
Table 12
Table 13Super-Bond Series
List of Major Components
Physical Properties of Polymerized Super-Bond
Impact Fracture Energy of Adhesives
Durability of Bond to Precious Metals using V-PRIMER
Water Sorption and Solubility in Water of Various Luting Cements
Bond Strength to Tooth Substance
Bond Strength to Metals
Bond Strength to Porcelain using Porcelain Liner M
Effect of Polymer/Monomer on Working Time and Curing Time in Bulk-mix Technique
Polymer Types and their Curing Times in Brush-dip Technique
Effect of Temperature on Working TimeRadiopacity using Polymer (Bulk-mix Radiopaque)
Interaction of Porcelain Liner M and V-PRIMER
Effect of Heat Treatment Condition of Porcelain Liner M Coated Surface on Bond Strength
Effect of Dentin Treatment with Sodium Hypochlorite

Questions and Answers

Question 1
Question 2
Question 3
Question 4
Question 5
Question 6
Question 7
Question 8
Question 9
Question 10
Question 11
Question 12
Question 13
Question 14
Question 15
Question 16
Question 17

What is the adhesive monomer, "4-META"?
Is the mechanical strength of Super-Bond lower than other adhesive resins?
Does water absorption reduce Super-Bond's mechanical strength?
What is the "TBB" Catalyst?
How do I apply "Enamel Etchant Gel" and "Dentin Etchant Gel" to the enamel surface?
Why is the "Dentin Etchant Gel" used for treating dentin?
How do I treat a tooth surface that consists of both dentin and enamel?
What is the purpose of treating dentin with sodium hypochlorite? How should it be used?
How do I protect the surface of the prosthesis where I don't want the adhesive to bond?
How much working time does Super-Bond allow?
What is the difference between the Bulk-mix Polymer powders and the normal Polymer powders?
Can I use Dispensing Cups and Dispensing Stand at the temperature above 25˚C?
How long does the activated liquid remain active?
Do I have to pre-wet the surfaces being cemented with the activated liquid?
What are the different types of Polymer powders for? And how do I use them?
How can I remove excess resin?
What are the different types of Brush Tips for? And how do I use them?

PAGE
2
2
2
4
4
4
5
5
6
6
8
8
20

10
10
10
11
11
11
11
11
11
12
12
12
13
13
13
13

14
14
14
14
14
15
15
15
16
16
16
16
17
17
17
18
18

1

CAUTION

qAvoid contact
Avoid contact with soft tissue, skin or eyes. A rubber dam is recommended for intraoral use. Dentist should
use rubber or PVC dental gloves. Contaminated skin or mucosa should be wiped off immediately with alcoholand then thoroughly rinsed with running water, otherwise symptoms such as swelling may show. If
Super-Bond enters the eye, immediately rinse thoroughly with running water. The patient should be examinedby an ophthalmologist. When the cured adhesive contacts soft tissue, polish the adhesive surface.
wBe careful of acidity
As the Enamel Etchant Gel and Dentin Etchant Gel are acid, avoid contact with soft tissue, skin or eyes, anddo not allow patients to swallow them while applying and washing them with water.
eBe careful of flammability
Catalyst V and Monomer are flammable. Do not store where they may be exposed to open flame.
rClean spilled Catalyst V immediately with wet towels
The Catalyst V reacts with oxygen. If absorbed by a flammable material, it may raise the temperature enoughto cause smoldering. If the Catalyst V is spilled, wipe it up immediately with a WET(not dry) disposable towel.
Then rinse the towel to kill the catalyst thoroughly in running water.

1.What is Super-Bond?
Super-Bond is a self-cure dental adhesive resin cement based on MMA. It contains a high performancebonding monomer, "4-META"*1, and a catalyst, "TBB"*2. It shows excellent bond strength to tooth (enameland dentin), metal*3, porcelain*4 and dental resins.
For three decades it has been used extensively in a wide range of dental applications. During this long
period it has earned an excellent reputation for pulpal safety.
Super-Bond is widely known for its formation of a sound hybrid layer (resin impregnated layer) in bothenamel and dentin. This layer reinforces the tooth surface against recurrent caries and prevents the postoperative hypersensitivity.
*1 See Table 2 and Question 1.
*2 See Table 2 and Question 4.
*3 Use V-PRIMER concurrently for precious metal alloys.
*4 Use Porcelain Liner M concurrently for porcelain.

Brush-dip kit
qQuick Monomer ------------------------------------------ 10mL

uBrush Handle (Bent) ------------------------------------------ 1

wCatalyst V ------------------------------------------------- 0.7mL

iBrush Tips (Pink • L • Brush-dip) ------------------------- 10

ePolymer (Brush-dip Clear)--------------------------------- 3g

oBrush Tips (Purple • LL • Brush-dip) -------------------- 10

rEnamel Etchant Gel--------------------------------------- 3mL!0Needle Tips (23G) --------------------------------------------- 5

tDispensing Stand ---------------------------------------------- 1!1Needle Cap (Red) -------------------------------------------- 1

yDispensing Cups--------------------------------------------- 20
Attached documents:
Instructions / Pictorial Instruction Card

2.Contents of Super-Bond kits

C&B kitqQuick Monomer ------------------------------------------ 10mL
wCatalyst V ------------------------------------------------- 0.7mL
ePolymer (Brush-dip Clear)--------------------------------- 3grPolymer (Bulk-mix Radiopaque) ------------------------- 5g
tEnamel Etchant Gel--------------------------------------- 3mL
yDentin Etchant Gel ---------------------------------------- 3mL
uDispensing Stand ---------------------------------------------- 1
iDispensing Cups--------------------------------------------- 20

2

Attached documents:
Instructions / Pictorial Instruction Cards

oMeasuring Spoon (Standard) ------------------------------- 1!0Brush Handle (Bent) ------------------------------------------ 1!1Brush Tips (Blue • Bulk-mix) ------------------------------- 10!2Brush Tips (Pink • L • Brush-dip) -------------------------- 10!3Brush Tips (Purple • LL • Brush-dip) ---------------------- 10!4Needle Tips (23G) ------------------------------------------ 5x2!5Needle Cap (Red) -------------------------------------------- 1!6Needle Cap (Green) ------------------------------------------ 1

Bulk-mix kit
qQuick Monomer ------------------------------------------ 10mL

uDispensing Cups--------------------------------------------- 20

wCatalyst V ------------------------------------------------- 0.7mL

iMeasuring Spoon (Standard) ------------------------------- 1

ePolymer (Bulk-mix Esthetic)------------------------------- 3g

oBrush Handle (Bent) ------------------------------------------ 1

rPolymer (Bulk-mix Radiopaque) ------------------------- 5g!0Brush Tips (Blue • Bulk-mix) --------------------------- 10×2

tDentin Etchant Gel ---------------------------------------- 3mL!1Needle Tips (23G) --------------------------------------------- 5

yDispensing Stand ---------------------------------------------- 1!2Needle Cap (Green) ------------------------------------------ 1

Attached documents:
Instructions / Pictorial Instruction Card

3

3.Precautions

Read all instructions thoroughly before use.
3-1 Safety

Please keep the following precautions for safe use.
(Regarding Catalyst V, read 3-2 additionally.)
1 Applications
Use Super-Bond only for the applications recommended in this publication.
2 Past history of sensitivity
Super-Bond should not be used by clinicians or on patients who are methacrylic monomersensitive.
3 Symptomatic irritation
Cease using Super-Bond immediately, if signs of irritation such as rashes appear, and see aphysician.
4 Avoid contact
Avoid contact with soft tissue, skin or eyes. A rubber dam is recommended for intraoral use.
Dentist should use rubber or PVC dental gloves. Contaminated skin or mucosa should be
wiped off immediately with alcohol and then thoroughly rinsed with running water, otherwise
symptoms such as swelling may appear. If Super-Bond enters the eye, immediately rinse
thoroughly with running water. The patient should be examined by an ophthalmologist. Whenthe cured adhesive contacts soft tissue, polish the adhesive surface.
5 Be careful of acidity
As the Enamel Etchant Gel and Dentin Etchant Gel are acid, avoid contact with soft tissue,
skin or eyes, and do not allow patients swallow them during application or rinsing.
6 Pulp protection
If the preparation approaches the pulp, apply a protective base.
7 Give care to flammability
Catalyst V and Monomer are flammable. Do not store where they may be exposed to open flame.

3-2 Precautions on Catalyst V

Catalyst V reacts with air and water to generate heat and lose activity. Please abideby the following.
1 Storage conditions

Avoid high temperature, high humidity and direct sunlight. The Catalyst should NOT be
refrigerated. (The repeated temperature changes may shorten the Catalyst's shelf-life by causing
the syringe to aspirate air.)*After a long storage, the first drop of the Catalyst may be inactive, though the rest of thematerial remains active.
*The syringe is made of glass, therefore it must be handled with care to prevent shock, dropping,and other physical damage.

2 Cap closure

The cap simply slides on and off. Recap the syringe immediately after each use. Air (oxygenand humidity) deactivates the Catalyst. Do not leave the cap off during the bonding procedure.

3 Screwing

If the Catalyst does not come out of the syringe because of the tight screw, do not try too
hard to trun it. The content may splash as the syringe breaks.

3-3 Storage

Please take the following precautions to maintain the quality.
1 Storage conditions
As in the case of the Catalyst V, store the Monomer, Polymer, Enamel Etchant Gel and Dentin
Etchant Gel in a cool, dark location. High temperature, high humidity and direct sunlight will
shorten their shelf-life.
2 Volatility
Monomer is highly volatile. Recap the bottle immediately.
3 Contamination
Do not mix the bottle caps.
4 Dispensing Cups and Brush Tips disposal
Both Dispensing Cups and Brush Tips are for one-time use. Dispose of them after use.

3-4 To Get the Best Results with Super-Bond
1 Create and maintain a clean surface
Oil, blood, saliva and biofilm will lower the bond strength. Clean the tooth and prosthesis
thoroughly before cementing. After cleaning, take care to avoid re-contamination.
2 Dry the surfaces and prevent moisture contamination
After cleaning, dry the surface adequately. A rubber dam is highly recommended, as it will
reduce the chance of contamination by saliva, humid breath or blood.
3 Avoid eugenol-containing bases and cements
Eugenol is a polymerization inhibitor. Therefore, eugenol-contained bases and cements should
not be used with resin cements. To avoid cross-contamination, reserve a mixing dish exclusively
for Super-Bond. Do not use the same mixing dish for other adhesives.
4 Time constraints
Super-Bond's working and setting times are very different from those of traditional cements.
Follow the instructions carefully to get the best results.
5 Do not re-use Polymer
After using Super-Bond in the Brush-dip technique, dispose of any excess Polymer left in the
mixing dish. Do not return it into the container, as it has become contaminated with the
Monomer.
6 Prosthesis Design
To avoid stress concentration which encourages debonding regardless of the actual bondstrength, design a prosthesis, such as wings of a bonded bridge, without thin unsupported area,
which may flex during mastication.
As in any dental treatment, the patient's individual constitution and the unique requirementsof clinical case at hand must be considered before selecting materials and conditions foruse.

4 After use

Unscrew the male-screw two turns counter-clockwise after each use to relieve pressure on
the Catalyst. (Pressure buildup can cause leakage of the Catalyst or a crack of the syringe.)

5 Clean spilled Catalyst immediately with wet towels

The Catalyst reacts with oxygen. If absorbed by a flammable material, it may raise thetemperature enough to cause smoldering. If the Catalyst is spilled, wipe it up immediately
with a WET(not dry) disposable towel. Then rinse the towel to kill the Catalyst thoroughly
in running water.

6 Cleaning of the tip of the syringe

Wipe the tip of the syringe with a dry gauze after each use to prevent residue buildup. Thenrinse the gauze with water to kill any remaining activity. Buildup of the residue may preventthe cap from seating properly.

45

4 How to use Super-Bond
Super-Bond can be used either with the Bulk-mix technique or the Brush-dip technique.
Choose the appropriate technique by referring to the table below.

Comparison of Techniques

Outline oftechniqueComparisonof the twotechniques

Bulk-mix technique

Brush-dip technique

Polymer powder is mixed directly to theactivated liquid*.

The powder/liquid ball is formed at the tip of abrush by dipping the tip first into the activatedliquid* and then touching the Polymer powder.

Use the powder/liquid mixture immediately.

The activated liquid must be used up within 5
minutes.

Applicable to comparatively wide area.

Applicable to comparatively narrow area only.

As the powder/liquid ratio is lower than thatof Brush-dip technique, the working time iscomparatively long but the curing is slow.

As the powder/liquid ratio is higher than that of
Bulk-mix technique, the working time of mixed ballis comparatively short and the curing is fast.

*Mixture of 4 drops of Monomer and 1 drop of Catalyst V

Surface Preparation

Porcelain Surface
Cleaning

Washing and drying

Application of Porcelain Liner M

Heat treatment

Metal Surface
Cleaning

Sandblasting

Protection of adjacent surfaces

Application of V-PRIMER

It is essential that all surfaces to be bonded with Super-Bond should be properly prepared. Preparation
varies depending on the nature of the materials.

Tooth Surface
Moisture control

Cleaning

Isolation by rubber dam or cotton roll is recommended.
*Remove scales beforehandRemove contaminants and stains using a polishingbrush and oil-free, fluoride-free pumice. Rinse
thoroughly and dry.

Surface treatment

Washing and drying

Apply the appropriate Etchant directly from thesyringe.

Treatment time
Etchant
DentinEnamel6

Dentin Etchant Gel
5-10 sec
30-60 sec

Enamel Etchant Gel
30 secRinse thoroughly with water and dry.

Do not use Enamel Etchant Gel on dentin.
Alternately, enamel may be prepared with Dentin Etchant Gelfor 30-60 seconds.
(See Questions 5-7.)

7

4 How to use Super-Bond
Operation Steps (Contd.)

BULK-MIX TECHNIQUE

Preparation of Dispensing Stand

Place the Dispensing Cups

Preparation of the Activated Liquid

Dispensing the Monomer

Dispensing the Catalyst V

Mixing the Polymer

Application of the Adhesive

Seating the Restoration

Post Treatment

Insert the restoration immediately. After confirming that it
is completely seated, hold in
position until the cement sets.

Remove the excess cement.
To facilitate this, protect beforehand unbonded surface
properly and remove the excess resin timely.
(See Question 16.)

*The curing time varies withtemperature and the type ofPolymer. (See Table 8.)

Hold the Monomer bottle
vertical, and dispense theproper number of drops intoa Dispensing Cup.

Hold the Catalyst V syringe
vertical, and turn the screw todispense the proper numberof drops to the Monomer. Stirlightly with a brush. This
mixture is called "activatedliquid"(See Question 13.)

Using the supplied measuring
spoon, add the Polymer powder
to the activated liquid. Stir lightly
with a brush.
(See Question 15.)

Immediately after mixing, usea brush to apply the cement
to the surface being bonded.
(See Question 14.)

Key points to achieve good seating

Work quickly, and seat the restoration before the mixture begins to gel.

Mixing ratio

Monomer

Catalyst V

4 drops
8 drops

1 The recommended temperature while using Super-Bond is below 25˚C.

Polymer

1 drop1 small cup of Measuring Spoon

2 drops

1 Iarge cup of Measuring Spoon

2 Mix Super-Bond at the last moment before bonding.
3 If the temperature exceeds 25˚C, use the pre-cooled ceramic Dispensing Dish.

BRUSH-DIP TECHNIQUE
Dispensing the Polymer

Preparation of the Activated Liquid

Dispensing the Monomer

Dispensing the Catalyst V

Application of the Activated Liquid
Brush the liquid onto thesurface to be bonded.
(See Question 14.)

Dipping the brush

Brush-dip Procedure
Forming the ball*The activated liquid decomposes gradually and loses
activity.
Use it within 5 min. after preparation.

Dispense the appropriate
amount of Polymer powder
into a Dispensing Cup.

Mixing ratio

Monomer

4 drops
8 drops

8

Hold the Monomer bottle
vertical and dispense theappropriate number of drops
into another Dispensing Cup.

Hold the Catalyst V syringe
vertical, and turn the screw todispense the proper numberof drops into the Monomer.
Stir lightly with a brush. This
mixture is called "activatedliquid"(See Question 13.)

Dip the Brush Tip (forBrush-dip) into the activatedliquid.
*When you repeat the procedure,
clean up the brush with gauze
before you dip it.

Touch the brush to thePolymer powder in the
Dispensing Cup.
A small ball of powder will bepicked up on the wet tip of thebrush.

Applying the ball
Brush the powderball onto the prewet surface being
bonded. As soon
as it touches thesurface, the powder
will spread out tocreate a creamy,
homogeneous layer.
If necessary, repeat
the procedure until
the entire surface iscovered with thecement.

Seating the Restoration

Post Treatment

Insert the restoration
immediately. Afterconfirming that it iscompletely seated,hold in position until
the cement sets.

Remove the excess
cement.
To facilitate this,protect any adjacentsurfaces you will not
be bonding, andremove the excess
resin before it sets.
(See Question 16.)

Catalyst V
1 drop2 drops9

Data and References
Fig. 1 : Impact Fracture Energy of Adhesives

Dental adhesive resin cement

Method of use

Bulk-mix technique and Brush-dip technique

Characteristics(1)Self-curing dental adhesive cement basedon MMA,"4-META" (bonding monomer) and"TBB" (polymerization catalyst).
(2)Excellent bond strength to dentin, enamel,
metal, porcelain and resins for dental use.
(3)Super-Bond forms "Hybrid layer" withdentin. This layer produces q protection
against recurrent caries and w isolation of
the pulp from outer stimuli.

V-PRIMER

Product nameMain use
Method of use
Characteristics

Porcelain Liner M

Adhesive primer for precious metal alloys

Adhesive primer for porcelain(1)One component adhesive primer for precious metal alloys.
(2)Based on "VTD", a derivative of triazine dithiol.
(3)A single coat of V-PRIMER improves remarkably the durability of Super-Bond to
precious metal alloys.
This primer eliminates the need for otherbond-enhancing steps, such as heat treatment
or tin plating.

(1)Two-component adhesive primer for porcelain.
(2)Application of Porcelain Liner M improves
remarkably the bond strength of Super-Bond
to porcelain and its durability.

Single liquid application

Components

Major Constituents

0.7mL TBB, Hydrocarbon
10mL MMA, 4-META

Quick Monomer* 10mL
Polymer
Clear
3g
Esthetic
3g
Opaque Ivory
3g
Opaque Pink
3g
Bulk-mix Clear
3g
Bulk-mix Esthetic
3g
Bulk-mix Radiopaque 5g
Brush-dip Clear
3g
Enamel Etchant Gel 3mL
Red Activator
5mL
Dentin Etchant Gel 3mL
Green Activator
5mL

MMA, 4-META

Application of the mixture of two liquids

In the Kit
In all kits
Optional

In all kits

Optional
PMMA
Optional
PMMA, pigments
Optional
PMMA, pigments
Optional
PMMA, pigments
Optional
PMMA
PMMA, pigments
Bulk-mix
PMMA, radiopaque pigments C&B, Bulk-mix
PMMA
C&B, Brush-dip
C&B, Brush-dip
Phosphoric acidOptional
C&B,
Bulk-mix
Citric acid, FeCl3
Optional*Super-Bond Quick Monomer is a monomer with curing time faster thanSuper-Bond Monomer. Other features, such as working time and bondstrength, are essentially the same as Super-Bond Monomer.

Items
Compression Strength[Elastic deformation limit]
Flexural Strength[Elastic deformation limit]

ValuesMeasured based on

84MPa

JIS T6602

67MPa

ISO4049

1000

600
400

Fig. 2 : Durability of Bond to Precious Metals
using V-PRIMER

200

A

B

C

31µg/mm

JIS T6514

Film Thickness
Standard Polymer/Monomer
ratio Polymer reduced to 2/3of the above

20-30µm
15-20µm

3

12µg/mm3

JIS T6602
JIS T6602

Water Sorption Solubility in Water(µg/mm3)(µg/mm3)

EC (Zinc phosphate cement)
HC (Carboxylate cement)
FB (Glass ionomer cement)
BR (Resin cement)
ID (Resin cement)
PT (Resin cement)
Super-Bond

419.3
309.3
211.6
24.2
31.5
32.2
31.2

41.3
33.8
34.4
14.2
9.5
17.8
12.1

Table 5 : Bond Strength to Tooth Substance

DentinSurface Treatment
Enamel Etchant Gel
Dentin Etchant Gel
Dentin Etchant Gel

Bond
Strength
(MPa)
15
13
17

Table 6 : Bond Strength to Metals

MetalGold alloy Type IV

Bond
Surface Treatment Strength
after Sandblasting(MPa)
V-PRIMER
400˚C, 5min.
Tin-plating

V-PRIMER
Gold/Silver/Palladium alloy 400˚C, 5min.
Tin-plating
Nickel-Chromium
Cobalt-Chromium
Hardened amalgam

10

F

30

V-PRIMER
Tin plating
Heat treatment
None

20
10

20,00050,000100,000

Thermal cycle (Times)

Table 4 : Water Sorption and Solubility in Water of
Various Luting Cements

EnamelWater Absorption

11

E

0
(MPa)

ISO10477

ISO4049

2

D

Adhesives
B : Phosphate ester contained
A : Adhesive wax
E : Carboxylate
C,D : 4-META contained
F : Ionic polymer(Adhered material is Crystalline Glass of Calcium Phosphate)*quoted from the Quintessence Vol.4, No.2, 1985

JIS Z2243

18X10 MPa

Solubility

1200

Tooth
Substance

Modulus of Flexural Elasticity
Brinell Hardness

1400

CementTable 3 : Physical Properties of Polymerized Super-Bond

Table 2 : List of Major Components

Components
Catalyst V
Monomer

Impact fracture energy ( x103

Main use

dyne • cm )Super-Bond

Product nameBond strength

Table 1 : Super-Bond Series

-

28
25
23
28
24
22
30
31
10

Table 7 : Bond Strength to Porcelainusing Porcelain Liner M
1. Bond Strength between VITA Porcelain(ground with #600 Emery paper) and Stainless Steel *1
Thermal Cycle (Times)(4˚C-60˚C)

Bond Strength *3 (MPa)
1,000

5,000

10,000Porcelain Liner M,
Super-Bond

20

19

13

A bonding agent forporcelain (Imported)

20

12

4

Super-Bond
without
Porcelain Liner M

8

-

-

2. Bond Strength between VITA Porcelain(glazed surface)and Stainless Steel *2
Thermal Cycle (Times)(4˚C-60˚C)

Bond Strength *3 (MPa)
1,000

5,000

10,000Porcelain Liner M,
Super-Bond

20

19

9

A bonding agent forporcelain(Imported)

20

5

3

3. Bond Strength between Ceramic Bracket and Acrylic Block
Thermal Cycle (Times)(4˚C-60˚C)

Porcelain Liner M,
Super-Bond

Bond Strength *3 (MPa)
1,000

5,000

10,00020

1110Notes: *1 Bonding of VITA(559)#600 to SUS304(Sandblasted)*2 Bonding of the glazed surface of VITA(559) to
SUS304(Sandblasted)*3 Tensile bond strength after completion of thermal cycle

11Data and References
Table 8 : Effect of Polymer/Monomer on Working Time and Curing Time in Bulk-mix Technique

Polymer

Working Time (23˚C)*1(sec.)Monomer

Clear
Bulk-mix
Esthetic
Type
Radiopaque

Quick
Monomer

Working Time (16˚C)*1(sec.)MonomerCuring Time (37˚C)*2(min.)Monomer

Quick
Monomer

120

13

8

150

14

9

70

12.5

6

110

13.5

8

Clear

NormalType

QuickMonomer

Esthetic

Opaque Ivory
Opaque Pink

*1 Available time before threading starts (namely, in slurry or sol state) at 23℃/16℃*2 Time to wait before occlusion

Brush-dip
Clear
Type

Quick
Monomer

10

5

Clear
NormalType

Esthetic

Opaque Ivory

180
120

*1 Radiopacity of Aluminium is regarded as 100%
(Test Method is based on ISO 4049)

Table 11 : Interaction of Porcelain Liner M and V-PRIMER

Primary
Coating

Bond
Strength(MPa)*

Secondary
Coating

V-PRIMER

Porcelain Liner M

25

V-PRIMER

V-PRIMER

Porcelain Liner M

Porcelain Liner M

V-PRIMER

Porcelain Liner M

24
25
22
12
16

Porcelain Liner M*Composite resin is bonded with Super-Bond after coating of the Primer(s) and bond strength is measured after 1,000 times of thermal cycles

Table 12 : Effect of Heat Treatment Condition of
Porcelain Liner M Coated Surface on Bond Strength

Heating Condition

Treatment Bond
Time
Strength(min.)(MPa)

Not heat treatedFig. 3 : Effect of Temperature on Working Time

400

Bulk-mix Clear/Esthetic
Bulk-mix Radiopaque
Clear/Esthetic
Opaque Ivory/Opaque Pink

350
Working Time (sec.)

210

DentinV-PRIMER

7

Opaque Pink

1

EnamelPorcelain

5

11Super-Bond using
Polymer Bulk-mix RadiopaqueGold/Silver/Palladium
alloy

Curing Time (37˚C)(min.)Monomer

Ratio to Standard
Radiopacity*1
Polymer/Monomer Ratio(%)

Material

Base
Material

Table 9 : Polymer Types and their Curing Times in Brush-dip Technique

PolymerTable 10 : Radiopacity using Polymer (Bulk-mix Radiopaque)

300
250

Heat treated with a dental
blower (120-130˚C)
Heat treated with a blow
dryer (70-80˚C)

13
1
2
1
3

19
20
17
21

Table 13 : Effect of Dentin Treatment
with Sodium Hypochlorite (10% solution)

Treatment Time with Sodium
Hypochlorite (sec.)

Bond
Strength(MPa)*1

0

17

15

16

30

13

60

6

*1: The dentin surface is treated with sodium hypochlorite and then with
the Dentin Etchant Gel, and bonded with Super-Bond

200
150
100
50
0

12

16˚C

23˚C

25˚C

27˚C

30˚C

13

QUESTIONS and ANSWERS
Question 1 : What is the adhesive monomer, "4-META" ?

Question 6 : Why is the "Dentin Etchant Gel" used for treating dentin?

Answer : "4-META" is an abbreviation of "4-methacryloxyethyl trimellitate anhydride." It has the structure shown below.
4-META is a derivative of MMA and is polymerized with MMA into a co-polymer. It contributes to the excellent
bonding property of Super-Bond.

Answer : Super-Bond bonds to dentin through the formation of a resin-impregnated layer.
For the best resin penetration, the surface smear layer should be removed without excessive modification of theunderlying dentin.
The "Dentin Etchant Gel" is ideal for treating dentin because it dissolves less hydroxyapatite, and minimizes
damage to dentinal collagen. Treatment for 5-10 seconds with the "Dentin Etchant Gel" is adequate.
"Enamel Etchant Gel", on the other hand, tends to excessively decalcify the dentinal hydroxyapatite and denature
the remaining collagen. This is not conducive to formation of a stable resin-impregnated layer, so bond strength
deteriorates.

Question 2 : Is the mechanical strength of Super-Bond lower than other adhesive resins?
Answer : Since the composition of Super-Bond is different from that of composite resin type adhesive with high load ofinorganic fillers, it shows lower compressive strength and hardness, but less brittleness.
It permits a tougher, more flexible bonding layer that can disperse stress and help the prosthesis endure impact
and torsion without loosening or fracturing. In fact, Super-Bond's unique flexibility due to its nature is an
important factor contributing to its superb clinical reputation.

Question 3 : Does water absorption reduce Super-Bond's mechanical strength?
Answer : Some reports emphasize the water absorbency of Super-Bond and suggest that this may affect its durability.
Because it consists almost exclusively of PMMA, which contains a hydrophilic group but is basically a watertightresin. Water sorption and water solubility values of Super-Bond are as low as those of other resin cements.
These values are almost negligible, if compared with inorganic cements such as zinc phosphate cements,
carboxylate cements or glass ionomer cements. (See Table 4.)
According to a clinical report, Super-Bond exposed at the crown margin has been maintained for more than 10
years without color change. After a full decade in the mouth, the cement exhibited excellent margin seal with
only minor surface abrasion.
When applied under normal seating conditions, Super-Bond prevents recurrent caries and post-operative
sensitivity often observed with conventional cements and generally ascribed to cement washout. Crowns
cemented with Super-Bond enjoy an excellent clinical prognosis.

Question 4 : What is the "TBB" Catalyst?
Answer : Super-Bond's "TBB" catalyst is crucial to the adhesive's remarkable performance.
Pure TBB (tri-n-butylborane), an organic boron component, is so reactive that it can be hazardous to use in aclinical situation. So to allow safe handling, the excessive reactivity has been reduced in the Catalyst V throughpartial oxidation and addition of diluents.
This modification does not reduce its effectiveness as a catalyst. The Catalyst reacts with oxygen in the air andwater, and oxidizes into a peroxide. The peroxide further decomposes, forming radicals which initiate thepolymerization of the MMA.
The key to the clinical success with TBB is that the reaction proceeds in the presence of oxygen and water (bothof which are present in the tooth surface.) However, care should be taken when dispensing the catalyst, as
unnecessary contact with air will cause gradual decomposition and degradation.

Question 5 : How do I apply “Enamel Etchant Gel” and “Dentin Etchant Gel” to the enamel surface?
Answer : When enamel is treated with acid, it becomes decalcified. An irregular scale-like surface composed ofdemineralized enamel rods is formed.
Super-Bond's excellent wetting properties and superb penetration into the interprismatic surface combine toform a tenacious enamel bond.
The phosphoric acid in the "Enamel Etchant Gel" is a stronger decalcifier than the citric acid in the "Dentin Etchant Gel".
So we recommend the following selection criteria.
"Enamel Etchant Gel": For enamel surfaces without tooth reduction. The "Enamel Etchant Gel" is particularly
appropriate if the enamel has been treated with fluoride.
For enamel surfaces with shallow tooth reduction which has not exposed the dentin.
"Dentin Etchant Gel": For tooth surfaces in which both enamel and dentin are exposed.

14

Question 7 : How do I treat a tooth surface that consists of both dentin and enamel?
Answer : If the areas involved are small, it may be virtually impossible to properly apply the two different etchants for two
different periods to their respective surfaces.
In these cases we recommend applying "Dentin Etchant Gel" for 10 to 30 seconds, depending on the relative
size of the enamel and dentin surfaces (the more enamel involved, the longer the application.)
When enamel surrounds a cavity where the preparation breaks the DEJ, first apply "Dentin Etchant Gel" to theenamel. After 30 seconds, apply "Dentin Etchant Gel" to the dentin and allow it to remain for just 5 seconds.
Then rinse the tooth with water.

Question 8 : What is the purpose of treating dentin with sodium hypochlorite (NaOCI)? How should it be used?
Answer : a.To increase bond strength?
For some resin cements sodium hypochlorite is recommended to improve the bond strength to dentin. This is
NOT true with Super-Bond. In fact, if sodium hypochlorite is used after treating the dentin with the"Dentin Etchant Gel", the bond strength will be significantly reduced. Sodium hypochlorite should NOT be
used after the dentin has been prepared with the Dentin Etchant Gel.
b.For endodontics
A diluted aq. solution of sodium hypochlorite (below 10%) is often used during endodontic therapy to dissolveorganic materials, to decontaminate surfaces, or to stanch bleeding.
When using Super-Bond on surfaces treated with sodium hypochlorite, (or acidic electrolyzed water), thefollowing care should be taken to avoid compromising Super-Bond's exceptional bond strength.
1.Do NOT apply sodium hypochlorite after acid etching. Do NOT use sodium hypochlorite higher than 10% concentration.
Sodium hypochlorite will dissolve the collagen exposed by acid treatment. Therefore, the Dentin Etchant Gel
should be applied AFTER treatment with sodium hypochlorite.
2.Limit the treatment time with sodium hypochlorite to less than 30 seconds.
Short-term treatment does not significantly affect the bond. Prolonged treatment with sodium hypochlorite, however,will significantly decrease the bond strength:<Super-Bond bonded to dentin after treatment with 10% solution of sodium hypochlorite for various time periods>
Treatment time with sodium hypochlorite Seconds 0 15 30 60
Tensile Bond StrengthMPa
17 16 13 6
3. If NaOCl treatment time exceeds 30 seconds, neutralize the surface with a reducing agent before etching.
Method 1:
Apply Saforide RCTM, containing 38mg/mL diamine silver fluoride ( Ag(NH3)2F ), for a length of time 1/2nd as longas the surface was treated with sodium hypochlorite. Rinse with water and dry. Treat with the Dentin Etchant
Gel for 10 seconds. Rinse and dry. Then apply Super-Bond according to the ordinary procedures.
<The surface was treated for 60 seconds with a 10% solution of sodium hypochlorite, neutralized with Saforide RC , andthen bonded with Super-Bond.>
Treatment time with Saforide RCTM Seconds 30 60 Caution: Saforide RC may darken tooth surfaces.
TM

Tensile Bond StrengthMPa

8

13 *Saforide RCTM is a product of Bee-brand Medico-dental.

15

QUESTIONS and ANSWERS
Method 2:
Prepare a 10% aq. solution of ascorbic acid (or its Na or K salt). Apply the solution for a length of time 1/3rd
as long as the surface was treated with sodium hypochlorite. Dry the surface. Treat with the Dentin Etchant
Gel for 10 seconds, rinse and dry. Then apply Super-Bond according to the ordinary procedures.
Treatment time with NaOCl
Treatment time with ascorbic acidTensile Bond Strength

Seconds
Seconds
MPa

180
30 60
7
16

300
50 100
6
17

Question 13 : How long does the activated liquid remain active?
Answer : In contact with air, the "Catalyst" gradually decomposes and degrades.
Therefore, it is important to use the activated liquid (the mixture of Monomer and Catalyst V) as soon as
possible after mixing. For the Bulk-mix technique, you should complete all the preparations for bonding before
you begin preparing the activated liquid, then use it immediately.
For the Brush-dip technique, use the activated liquid within 5 minutes at the latest.

4. Hydrogen peroxide (H2O2)
In alternate irrigation, a hydrogen peroxide solution is sometimes used in connection with a sodium hypochlorite solution.
Hydrogen peroxide damages the dentin structure, thus compromises the bond strength, if treated for an extended
period beyond 30 seconds. Consequently, the use of a hydrogen peroxide solution should be limited to 30 seconds.
5. Acidic electrolyzed water
When used on dentin, acidic electrolyzed water reduces the bond strength of Super-Bond much like sodium
hypochlorite (though the strength reduction is less severe.) To achieve normal bonds after the tooth surface has beentreated with acidic electrolyzed water, apply the Dentin Etchant Gel for 30 seconds (instead of the normal 10
seconds). Rinse and dry. Then apply Super-Bond according to the ordinary procedures.

Question 9 : How do I protect the surface of the prosthesis where I don't want the adhesive to bond?
Answer : Before cementing an inlay, onlay, crown or bridge with Super-Bond, all exterior surfaces should be polished andthe interior surfaces sandblasted. Unfortunately, once it's allowed to cure, Super-Bond will adhere tenaciously
even to the highly polished exterior surfaces. Once it has completely polymerized, removing the excess with ahand scaler is extremely difficult.
To facilitate cleanup, the exterior surfaces may be coated with a separator including Super-Bond SEP by
Sun Medical.

Question 10 : How much working time does Super-Bond allow?
Answer : If you use Super-Bond in the Bulk-mix technique, the slurry-like cement gradually increases its viscosity andbegins threading.
Super-Bond must be used before threading begins. Otherwise, the operation becomes quite difficult. The
increase in film thickness may prevent seating of the prosthesis. (See Table 8, Fig. 3 and Question 11.)
*Unlike conventional cement, it is not necessary to spatulate Super-Bond. When using it for the first time, you
may feel the mixture is too thin. Be assured that Super-Bond should be used in this state.
*Super-Bond undergoes the following stages from mixing to curing. For proper seating of the prosthesis, try tofinish the bonding operation before the mixture has reached stage 3 (mild threading).
1. Slurry stage: The mixture is very thin, as when gypsum powder is mixed with a copious amount of water andthe powder particles are still visible.
2. Sol stage: The powder particles are no longer visible, but the mixture still retains high fluidity and has notstarted threading.
3. Mild threading: The mixture becomes sticky and forms thin threads.

Question 11 : What is the difference between the Bulk-mix Polymer powders and the normal Polymer powders?
Answer : Because of its particle size, Bulk-mix Polymer powders do not require chilling of Dispensing Stand andDispensing Cups as long as the temperature is below 25℃. On the other hand, the normal Polymer powders
require a cooled ceramic Dispensing Dish. The curing reaction of Super-Bond is influenced by temperature: The
higher the temperature, the more quickly Super-Bond cures. In a cooled Dispensing Dish, the curing reaction
proceeds slowly, the working time is extended, and the bonding operation becomes easier. (See Fig. 3.)

Question 12 : Can I use Dispensing Cups and Dispensing Stand at the temperature above 25˚C?
Answer : Dispensing Cups and Dispensing Stand are made of plastic and can not be chilled in a refrigerator. Instead, use
the ceramic Dispensing Dish, which is sold separately, for cooling.

16

Question 14 : Do I have to pre-wet the surfaces being cemented with the activated liquid?
Answer : Brushing the activated liquid onto the surfaces to be bonded before cementing will assure intimate resin contact.
In the Brush-dip technique, the ball has a comparatively high Polymer/Monomer ratio, so pre-wetting the surface
with the activated liquid is essential to assure maximum cement adaptation and to create the highest possible
bond strength.
In the Bulk-mix technique, application of the activated liquid is not necessary, so long as you seat the prosthesis
while the cement is in the slurry state. In this case, the mixture is fluid enough to assure maximum adaptation.
On the other hand, advanced-application of the activated liquid on the dentin surface is recommended torestrain the outflow of exudate from the surface of the prepared dentin and to prevent bacterial entry into thedentin tubules.

Question 15 : What are the different types of Polymer powders for? And how do I use them?
Answer : Eight different Polymers are available for Super-Bond, including those sold separately. They can be divided into
3 basic classes, the normal type, Bulk-mix Polymer and Brush-dip Polymer. Refer to the following list to select thebest Polymer for your application. (See Tables 2, 8 and 9 and Fig.3.)
1. Clear, Brush-dip Clear and Bulk-mix Clear
Fine PMMA powder without pigment. When cured, its medium translucency and inconspicuous shade is ideal
for temporary splinting of loose teeth, creation of temporary prostheses using a resin tooth or extracted tooth,or direct bonding of orthodontic brackets with the Brush-dip Clear. When the normal Clear polymer is used in
the Bulk-mix technique, the working time before threading is rather short. Here, the Bulk-mix Clear ispreferable, because it allows more working time.
2. Esthetic and Bulk-mix Esthetic
Tooth-colored Clear powder. When cured, its color is ivory with some translucency and little opacity. It is used
for prostheses that require an esthetic appearance. Curing time and workability are the same as Clear powder
and Bulk-mix Clear powder respectively.
3. Opaque Ivory
Opacity is produced by specially prepared pigments. The powder creates an extremely opaque cement film thatis highly effective in masking metal surfaces. It is recommended for use when metal show-through might affectaesthetics (as when cementing inlay or adhesion bridges), or when repairing fractured prostheses with exposed
metal. Its working time is slightly longer than Clear powder. The Opaque Ivory powder is widely used forroutine cementation. Due to its opacity, any excess cement is extremely visible, which facilitates removing it.
4. Opaque Pink
Pale pink color is added to Opaque Ivory. This powder was originally intended for repairing denture bases, but
it can be used for the same purposes as the Opaque Ivory. Some dentists prefer it to the Opaque Ivory as its
pink color reflected on the surrounding teeth produces a more natural shade. Its handling and setting properties
are similar to those of the Opaque Ivory.
5. Bulk-mix RadiopaqueThis powder contains highly radiopaque filler. When it is used under the standard Polymer/Monomer ratio, thecured adhesive shows radiopacity equivalent to enamel. It creates a natural tooth-color and allows a longer
working time for easier handling. However, it cures slightly more slowly than the Clear powder. For fastercuring, increase the Polymer ratio to Monomer.

17

QUESTIONS and ANSWERS
Question 16 : How can I remove excess resin?
Answer : When the object is seated, excess resin is expressed around the margins. If the excess resin is allowed to cure,it will bond to the surface of the prosthesis, and clean-up and polishing will be extremely difficult.
The following describes how to remove the excess resin:
qBefore cementing apply as much protection as possible to surfaces that should not be bonded. (See Question 9.)*Apply a separating agent on metal surfaces, wrap with "Parafilm," place retraction cord under the gingival
margin, etc.
wUse the Opaque Polymer, as the excess resin will be more visible.
eUse a cotton pledget or a brush soaked with alcohol to remove as much excess resin as possible immediately
after the prosthesis is seated (before threading begins). Resin extruded onto the interdental papilla should be
removed immediately using floss, etc.
Resin spilled onto the gingiva should also be removed before it hardens.
rUsing hand scalers, scrap off the residue after the threading stage is completed but before the cement iscompletely cured.
*Wait a while till the threading stage is completed.
*If you attempt to scrape off the excess during the threading stage, you may feel that you are removing all thecement, but a thin layer will remain bonded to the surface.
tIf the surface was coated with a separating agent, even cured resin can be easily peeled off using hand
instruments.
yAny remaining resin will be easily discovered by re-examination on the following day.

Question 17 : What are the different types of Brush Tips for? And how do I use them?
Answer : The Brush Tip Blue is a blunt-cut type and suitable for transportation of a large amount of the slurry in Bulk-mix
technique. The Brush Tips Green (S), Pink (L) and Purple (LL) have pinpointed tips and are suitable forBrush-dip technique. The Brush Tips Pink (L) and Purple (LL) are longer than Brush Tip Green (S) while thebristles of Brush Tip Purple (LL) are thicker than the others. Refills of the respective types are available.
Dispose of them after use.

18

19

5 Examples of Clinical Applications
CLINICAL APPLICATIONS using the BULK-MIX TECHNIQUE
Cementing of inlays,onlays and crowns

Salvaging fractured teeth

Bonded partial onlay crownOcclusion reconstruction using bonded onlay

Bonding fragments directly to the post

Extraction,cementation,and replantation

Cementing of posts and cores

Bonded prefabricated posts with adhesive amalgam core

Bonded cast post-and-core

Cementing of bridges
Anterior bonded bridge with crowns and retentive wings

Cementing of periodontal splints

after 9 years and 8 months

Adhesive splint made of mesh plate

Cast metal lingual splint

Bonded segmented bridge

after 7 years

after 13 years and 5 months

CLINICAL APPLICATIONS using the BRUSH-DIP TECHNIQUE
Temporary periodontal splinting of mobile teeth
By proximal bonding

Temporary prosthesis using a resin tooth or an extracted toothResin tooth

20

Resin tooth

Orthodontic treatment
Adhesion of metal bracket

Adhesion of porcelain and plastic bracket

Adhesion of retainer

Repair of fractured crown

Extracted tooth21齿科树脂粘接剂【形状、构造等】
本品构成品的组成成分见下表:
构成品

性状

组成

常规型单体液液体 MMA、4-META、其他

快速型单体液液体 MMA、4-META、其他

V型催化剂

液体 三正丁基硼部分氧化物、其他

聚合粉

粉末 PMMA、其他红色表面处理剂

液体 磷酸、水、其他绿色表面处理剂

液体 柠檬酸、三氯化铁、水、其他

高粘度红色表面处理剂 液体 磷酸、增粘剂、水、其他
附属品:调盘（陶瓷）、调盘台（3孔）、梅花杯、海绵（L、S）、
量勺（Standard、Small、Large）、毛刷柄（直柄）、毛刷柄（曲柄）、毛刷头（绿色·S）〈用于笔积法〉、毛刷头（粉色·L）〈用于笔积法〉、
毛刷头（紫色·LL〈用于笔积法〉、毛刷头（蓝色）〈用于混合法〉、
针头（23g），针帽（红色、绿色）[原理]
本品由单体液、聚合粉、聚合催化剂混合，在常温下聚合固化。

【适用范围】
本品适用于口腔治疗中牙本质、牙釉质、齿科用金属合金、陶瓷的粘接。
（Super-Bond Brush-dip套装不适用于牙本质的粘接。）

【性能】
项目

规格

试验方法

薄膜厚度

50μm以下ISO 4049:2000

固化时间

10分以内

ISO 4049:2000

粘接强度牛牙牙釉质※1 5MPa以上
牛牙牙本质※2 5MPa以上
金银钯合金※3 15MPa以上
陶瓷※4

10MPa以上

氧化锆※5

10MPa以上

自社法

挠曲强度

50MPa以上

X射线阻射性※6

同等或以上厚度的铝板 ISO 4049:2000

ISO 4049:2000※1 使用红色表面处理剂 ※2 使用绿色表面处理剂
※3 使用齿科贵金属粘接前处理剂【V-PRIMER】※4,5 使用齿科陶瓷粘接前处理剂【Porcelain Liner M】※6 仅对L型遮色聚合粉适用。

【操作方法或使用方法】
1.处理粘接面（1）牙面①按照常规方法，
对粘接牙面进行充分地清洁、水洗及干燥处理。
②若粘接牙面为牙釉质，可使用红色表面处理剂或高粘度红色
表面处理剂进行表面处理。也可使用绿色表面处理剂进行表面处理。
若粘接牙面为牙本质，可使用绿色表面处理剂进行处理。处理
时间请参考下表。
③处理后，请充分水洗并干燥粘接牙面。
红色表面处理剂或高粘度红色表面处理剂绿色表面处理剂牙釉质

30秒

30 ～ 60秒

牙本质

—

5 ～ 10秒

使用表面处理剂时，请遵照各型号的使用说明书进行操作。

使用说明书（2）金属面若粘接面为非贵金属，对粘接面进行氧化铝喷砂后，使用超声波进行清洗并干燥。若粘接面为贵金属，
先对粘接面进行喷砂处理，
然后用超声波进行清洗并干燥，
最后用齿科贵金属粘接前处理剂
（例如本公司的V-PRIMER）进行表面处理或镀锡、加热氧化处理。
使用齿科贵金属粘接前处理剂时，
请遵照其使用说明书进行操作。
（3）陶瓷（瓷材、氧化锆）面及树脂面
若粘接面为陶瓷，对粘接面进行必要的氧化铝喷砂处理，
并清洗、
干燥后，
然后用齿科陶瓷粘接前处理剂（例如本公司的Porcelain
Liner M）进行表面处理并用气枪轻轻吹干。若粘接面为氧化锆，对粘接面进行氧化铝喷砂处理，
并清洗、
干燥后，
然后用齿科陶瓷
粘接前处理剂（例如本公司的Porcelain Liner M）进行表面处理并用气枪轻轻吹干。若为瓷修补，对粘接面进行必要的研磨，去除
表面一层材料，并进行氧化铝喷砂处理，然后用超声波充分清洗并 干 燥 后，干 燥 后 用 齿 科 陶 瓷 粘 接 前 处 理 剂（例 如 本 公 司 的Porcelain Liner M）进行表面处理并用气枪轻轻吹干。
使用齿科陶瓷粘接前处理剂时，请遵照其使用说明书进行操作。
2.活性液的制备
在调盘（陶瓷）或调杯（配合调盘台（3孔）使用）中滴入4滴常规型单
体液或快速型单体液，再滴入一滴V型催化剂，使用毛刷头将混合液
搅拌2-3次后制备成活性液。5分钟内使用效果最好。
3.混合物（粘接剂）的调制、涂抹
在粘接操作中有以下两种方法（笔积法与混合法），
若进行衬里，
需在修复面上涂上薄薄的一层。
（1）笔积法将毛刷头装配到毛刷柄上，用毛刷头沾取的活性液，在粘接面上涂上一层。在调盘（陶瓷）或调杯（配合调盘台（3孔）使用）中放
入需要的聚合粉，用毛刷头沾取活性液，其带有活性液的毛刷头
尖与聚合粉末接触后制作水门汀小珠，然后涂抹于粘接面。
（2）混合法笔堆积法套装不适合用混合法操作。
本处混合法指的是C＆B套装。
将蓝色毛刷头装配到毛刷柄上，在活性液中加入黄色标准量勺
1.0勺聚合粉混合，用毛刷头沾取混合泥涂抹于粘接面。如需调
整固化时间，可调节聚合粉的使用量。
调盘（陶瓷）需放在冰箱中冷却后使用。
因聚合粉末的型号不同，操作时间（混合物状态）有所差异。
请参考下列表格，分开使用。
·使用了在冰箱中经过冷却的调盘（陶瓷）的操作时间
◇各聚合粉末 操作时间（混合法）
聚合粉

操作时间（10℃）
常规型单体液

L型透明色

约150秒

L型遮色

约200秒

（参考）使用了未冷却调盘（陶瓷）的操作时间
聚合粉
L型透明色、L型遮色操作时间（23℃）
常规型单体液约40秒

4.修复体的安装、压接、牙齿之间的固定修复体的安装、压接或牙齿之间的固定，必要时去除剩余的粘接剂。
粘接剂为半固化未拉丝的状态为去除剩余粘接剂的最佳时机。但是，若为聚合粉L型遮色时，安装、压接或固定后应立即使用干燥的棉球
擦掉修复体周围的剩余粘接剂。
5.树脂水门汀的固化
在修复体被压接固定的状态下，
使粘接剂固化。
固化基准:笔积法套装:笔堆积法固化时间5-6分钟。
C&B套装:L型透明粉笔堆积法固化时间7分钟，
混合法固化
时间8分30秒。L型遮色粉笔堆积法固化时间7分钟，
混合法
固化时间9分30秒。

固化时间有一定的基准，但实际上固化反应需要时间慢慢进行。因此，请患者在安装修复体之日不要咬过硬的物质。
6.抛光
待粘接剂完全固化后，
按照常规方法进行抛光。

②对本品或甲基丙烯酸酯类单体有过敏既往史的操作者，在操作时应佩戴牙科用手套，不应直接接触本品。特别是避免接触未固化的混合物。同时，因使用本品而引起过敏症状的情况，应立即停止使用，并接受专业医生的诊断。
③避免本品附着于口腔粘膜和皮肤，以及避免进入眼睛（建议使用橡皮障）。出现附着的情况，应立即使用酒精棉等材料擦拭，并用
大量的流水洗净。若进入眼睛，应立即用大量的流水洗净，必要时
接受眼科医生的诊断。
④使用部位的牙龈或黏膜若有伤或炎症时，请控制使用。
⑤使用本品时，应充分考虑患者的个体差异，以及判断是否属于本品的适应症后再使用。

[使用方法上的使用注意事项]1）前处理完毕后充分干燥粘接面，不要让唾液、呼气、血液等污染。
2）使用次氯酸钠溶液处理牙本质粘接面后，会使产品的粘接强度明显下降，请勿使用。有机成分的溶解、消毒、止血，使用时，在绿色表面处理剂处理前使用NeoCleaner※1，其处理时间为15秒以内。
在绿色表面处理剂处理后请勿使用NeoCleaner※1 ※1NEO制药工业（株）3）丁香酚物质（洞衬材料与水门汀）以及氟与油性研磨材会降低粘接3）不良、有害反应
性能，请勿使用。
使用本品的过程中，可能会出现发疹、皮炎等过敏症状。
※24）与HY 暂封材料并用时会产生着色现象。※2（株）松风5）氟强化的牙釉质，必须使用红色表面处理剂或高粘度红色表面处
【贮藏、保管方法】
理剂进行表面处理。
[贮藏、保管方法]
6）使用后应立即盖好盖子。
·常规型单体液、快速型单体液、V型催化剂严禁火源。
7）笔积法使用后的聚合粉不应装回原瓶中，应直接丢弃。
·本品应避免多湿、
阳光直射，存放于温度变化较小的室温场所（1℃8）使用后的调盘（陶瓷）里的残留物可使用纸巾等去除，并用专用的30℃）。V型催化剂不应置于冷库中保管。
洗液清洗。若已固化，请用水浸泡后，再清洗、干燥。
·本品应保管并管理在口腔行业者以外人员接触不到的场所。
【使用上的注意】
1）使用注意①请勿用于【使用目的、
功能或效果】以外的用途。
②无牙科医疗资格的人员请勿使用。
③操作者应戴好牙科用手套。
④请勿与其他产品混合使用。
⑤若窝洞较深，请对牙髓进行适当的保护。
⑥常规型单体液、快速型单体液、V型催化剂具有可燃性，请勿置于
火源处，或靠近火源处使用。
⑦吸入大量高浓度的MMA气体有可能会导致头痛等症状，
请在通风良
好的地方使用。若吸入大量高浓度的气体，应转移到新鲜空气处。
⑧使用、清洗的过程请勿误饮液体。若不小心误饮液体，请用水充分
漱口，必要时前往专科医生处就诊。
⑨V型催化剂与干燥的纸、纸巾、纱布、脱脂棉、海绵等易燃物接触后，有可能会着火。
·滴液时的注意
滴V型催化剂时，调盘（陶瓷）或调杯中不可放入海绵等易燃物，
否则可能会着火。
·溢出的V型催化剂的处理
不小心外溢或漏液时，必须使用用水湿润后的纸巾、纱布、脱脂棉等擦除。使用干燥的纸巾等擦除后，搁置在那边可能会着火。
若附着在衣物上，请立即用水清洗。
⑩V型催化剂的容器均为玻璃制品。有破损的可能性，应遵守以下注意事项：·跌落时撞击的注意跌落时产生的冲击力可能会导致容器损坏，
取用本品时应注意。
·挤压液体时的注意拧螺丝滴下V型催化剂时，因容器尖端的液体凝固而无法挤出时，不应用力拧动螺丝，否则有可能导致容器损坏，液体飞散。
·使用后的注意
使用后，将螺丝回旋两圈，让其无内压增加。内压上升后有可能
会导致漏液、容器损坏。
⑪V型催化剂与空气或水接触会发热分解，活性会大大降低。为保持
其性能请遵守以下注意事项:·盖帽的开闭
为避免由针尖接触到空气，
使用前打开盖帽，使用后应立即盖上。
·盖帽与针部的清洁盖帽与针部可能会附着上白色粉末。附着物会降低盖帽的密封性，所以请使用干燥的纱布等擦掉。使用后的纱布，请用水清洗。
⑫针头（23G）使用后请舍弃。
⑬使用后的容器应按[废弃物的相关法律]进行妥善处理。
2）重要的基本注意事项
①因使用本品而出现发疹、湿疹、发红、肿胀、瘙痒、皮疹、麻木等过
敏症状的患者，应立即停止使用，并接受专业医生的诊断。

【使用期限】·本品各构成品的有效期为3年。
·具体使用期限请参见包装标示，请在使用期限内使用本品。
【生产日期】
产品的生产日期请参见包装标示。
【说明书的修订日期】
2019年3月5日【包装】[套装]①Super-Bond C&B 套装·常规型单体液
1瓶（10mL）·V型催化剂
1支（0.7mL）
·聚合粉
L型透明色
1盒（3g）·聚合粉
L型遮色
1盒（5g）·红色表面处理剂 1瓶（5mL）·绿色表面处理剂 1瓶（5mL）·调盘（瓷器）1个·海绵球（大、小） 1盒
·量勺（标准）1把·量勺（小）
1把·毛刷柄（直柄） 1把·毛刷柄（弯柄） 1把·毛刷头（笔积S·白色） 1盒
·毛刷头（笔积L·白色） 1盒
·毛刷头（混合·青色） 2盒

②Super-Bond Brush-dip套装·快速型单体液
1瓶（10mL）·V型催化剂
1支（0.7mL）·聚合粉笔积用透明色1瓶（3g）·高粘度红色表面处理剂
1支（3mL 注射器）
·调盘台（3孔）1个·调杯20个
·毛刷柄（弯柄） 1把·毛刷头（笔积L·红色） 1盒
·毛刷头（笔积LL·紫色） 1盒
·针头（23G）
5个·针帽（红色）1个

注:以上内容物均有零售。
【产

地】

日本
【医疗器械注册证编号】国械注进20153170388【产品技术 要求编号】国械注进20153170388【注册人/生产企业】
名
称 : サンメディカル株式会社（尚美德齿科材料有限公司）
住
所 : 〒524-0044 滋贺县守山市古高町571-2
生产地址 : 〒524-0044 滋贺县守山市古高町571-2
电话 : +81-77-582-9978【代理人/售后服务单位】
名
称 : 日进齿科材料（昆山）有限公司
住所 : 江苏省昆山市高科技工业园北门路883号
电话 : 0512-57794412
J027C/1C5) Clean spilled Super-Bond Brush-dip Initiator immediately with wet towels
If absorbed by a combustible material, it may raise the temperature enough tocause smoldering. Spilled or unused Super-Bond Brush-dip Initiator should be
blotted up immediately with a WET (not dry) disposable towel. Then saturate thetowel to deactivate the Super-Bond Brush-dip Initiator thoroughly in running water.

EN FR DE I T ES TR

6) Problem in expressing the Super-Bond Brush-dip Initiator
If the Super-Bond Brush-dip Initiator does not form drop due to the blockage at theorifice or the screw does not turn, do not force it. The contents may splash if theglass syringe breaks. Please contact your distributor.

Super-Bond

Brush-dip
Starter Kit
Dental Adhesive Resin Cement

Super-Bond Brush-dip Starter Kit
FOR DENTIST USE ONLY
1. Product Overview
Super-Bond Brush-dip Starter Kit is a self-curing adhesive resin cement, it contains
4-META*1, MMA*2, TBB*3 and PMMA*4 (4-META / MMA-TBB resin). It may be used
to cement all fixed dental prosthesis fabricated from a wide range of dental
materials.
*14-methacryloyloxyethyl trimellitate anhydride*2Methyl methacrylate*3Tri-n-butyl borane*4Poly methyl methacrylate
Intended use:
Super-Bond Brush-dip Starter Kit is intended to adhere to tooth enamel, dentin,
metal, ceramic, zirconia and resin composite with the appropriate surface
pre-treatment.
Indications:Super-Bond Brush-dip Starter Kit is indicated for the followings:
a. Orthodontic applicationb. Direct fixation of mobile teeth
c. Direct bonded bridge
d. Repair of fractured prostheses

Contraindication:
Patient who has a history of severe allergic reactions to this product ormethacrylate monomers and acetone.
Side effect:
In rare cases, components of Super-Bond Brush-dip Starter Kit may lead tosensitization. In such cases, Super-Bond Brush-dip Starter Kit should no longer beused.
2. Components
Super-Bond Brush-dip Starter Kit* -------------------------- Class IIa
5[Class IIa]
Components

Amount or quantity in the kit
1 × 0.3 mL

1 Super-Bond Brush-dip Initiator

1 × 3.5 mL

2 Super-Bond Brush-dip Monomer

3 Super-Bond Polymer (Brush-dip Clear)1 × 1.5 g

4 Enamel Etchant Gel[Class I

1 × 1 mL]
Accessories

Amount or quantity in the kit

5 Dispensing Stand

1

6 Dispensing Cups

20

8 Brush Tips (Brush-dip L)

20

7 Brush Handle (Bent)

1
5

9 23G Needle Tip

Attached documents: Instructions, Pictorial Instruction Card, Precautions* Components of the kit may be purchased individually.
5

3. Compositions
1 Super-Bond Brush-dip Initiator:
Partially oxidized tri-n-butyl borane, Others.
2 Super-Bond Brush-dip Monomer:
Methacrylate monomers, Others.
3 Super-Bond Polymer (Brush-dip Clear):
Poly methyl methacrylate, Others.
4 Enamel Etchant Gel:
Phosphoric acid, Water, Thickener, Pigment Others.
Components

Form

1
2

Colorless transparent liquid

Colorless pale yellow transparent liquid

34White powder
Red gel

4. Precautions

4.1 Safety precautions

1 Super-Bond Brush-dip Starter Kit components contain substances, such as

methacrylate monomers, that may cause allergic reactions. They are not for use
on or by persons who are sensitive to acrylates, methacrylates or similar materials
Exposure to uncured resins may cause irritation, dermatitis or inflammatory
reactions to tissues (skin, mucosa, eyes, etc.) of sensitive individual. If this occurs,
discontinue the use, flush area with copious amounts of water and seek medical
attention, if necessary.
2 Avoid Super-Bond Brush-dip Starter Kit components contacts with soft tissues,
skin or eyes. Contaminated area should be wiped off immediately with alcohol andthen rinse thoroughly with copious amounts of running water. If the components
enter the eye, rinse immediately and thoroughly with copious amounts of runningwater and have the patient examined by an ophthalmologist.
3 Avoid accidental ingestion while applying and rinsing Super-Bond Brush-dip
Starter Kit components. When ingested, seek the medical attention, if necessary.
4 Handle Super-Bond Brush-dip Starter Kit components in a well ventilated area (airout the area several times per hour). In case of inhalation of the vapor, move forfresh air.
5 Super-Bond Brush-dip Initiator and Super-Bond Brush-dip Monomer are highly
volatile and flammable. Do not store where they may be exposed to open flame.
4.2 Common precautions
1 The use of rubber dam is recommended.
2 Dentists should wear dental gloves.
3 Do not mix components with those of other materials.
4 Use this product only for the applications recommended in these instructions.
4.3 Handling precautions

1 Contaminated surface with such as oil, blood, saliva and biofilms will lower thebond strength. Clean the tooth and prosthesis thoroughly. After cleaning, takecare to avoid re-contamination by isolating the surface adequately. Using rubber
dam is highly recommended.
2 Fluoride and oil will lower the bond strength. Dental paste containing fluoride or oilshould not be used for cleaning tooth surface.
3 Eugenol is a polymerization inhibitor. Therefore, eugenol-contained bases andcements should not be used with Super-Bond Brush-dip Starter Kit components.
4 Sodium hypochlorite is a polymerization inhibitor. When using Super-Bond
Brush-dip Starter Kit on the dentin surface treated with sodium hypochlorite,
Super-Bond C&B Green Activator or Dentin Etchant Gel should be applied after
sodium hypochlorite use. The treatment duration with sodium hypochlorite should
be limited to less than 15 seconds. Details on how to use Super-Bond C&B Green
Activator and Dentin Etchant Gel can be found in the attached document.
5 Super-Bond Brush-dip Initiator and Super-Bond Brush-dip Monomer are highly
volatile. Firmly recap immediately after use.
6 The Activated Liquid (mixture of Super-Bond Brush-dip Initiator and Super-Bond
Brush-dip Monomer) gradually deactivates. Use it within 5 minutes after mixing.
7 Super-Bond Brush-dip Starter Kit may be discolored, if combined with atemporary cement containing tannic acid.
8 Super-Bond Brush-dip Starter Kit components is single use only. Do not return
dispensed components to their original container. Dispose of leftover material
properly. After using Super-Bond Brush-dip Starter Kit in the Brush-dip technique,dispose of any excess Super-Bond Polymer (Brush-dip Clear) left in the dish.
9 Brush Tips and Dispensing Cups are single use only. Do not reuse. Discard after
each use.
10 When a preparation is deep, close to the pulp, protect the pulp with appropriatematerial.
11 Before using the accessories, check for damage, cracks, scratches, discoloration,etc., and stop using them if there is any abnormality.
4.4 Precautions on Super-Bond Brush-dip Initiator
When exposed to air or water, Super-Bond Brush-dip Initiator can lose activity andgenerate heat. Abide by the following.
1) Storage conditions
Avoid high temperature, high humidity and direct sunlight. The repeated change
in temperature may cause the syringe to aspirate air, thereby shortening theSuper-Bond Brush-dip Initiator’s shelf-life.
*After prolonged storage, the first drop of the Super-Bond Brush-dip Initiator may
have less activity due to the exposure to the air (oxygen), though the rest of thematerial in the container remains unaffected. Disposing of a drop before using
will ensure the proper chemical reaction.
*The syringe is made of glass, therefore it must be handled carefully to avoid
shock, dropping, and other physical damage.
2) After use
Turn the syringe screw two turns counter-clockwise after each use to relieve thepressure of the Super-Bond Brush-dip Initiator barrel. (Built up pressure cancause leakage of the Super-Bond Brush-dip Initiator or in some case crack thesyringe.)3) Cap closure
The cap simply slides on and off. Recap the syringe immediately after each use.
Air deactivates the Super-Bond Brush-dip Initiator. Do not leave the cap off.
4) Cleaning syringe
Wipe the tip of the syringe with a dry gauze after each use to prevent residue
building up. The gauze used should be saturated with water to fully deactivate
Super-Bond Brush-dip Initiator before disposal. Buildup of the residue may cause
the cap seat improperly.

Super-Bond Brush-dip Starter Kit
1. GénéralitésSuper-Bond Brush-dip Starter Kit est un ciment résine adhésif auto polymérisantqui contient du 4-META*1,MMA*2, TBB*3 et du PMMA*4 (4-META / MMA-TBB
résine). Il peut être utilisé pour coller tous types de prothèses dentaires fixes
fabriquées avec une large gamme de matériaux dentaires.
*1 : Anhydride de trimellitate de 4-méthacryloyloxyéthyle*2 : Méthacrylate de méthyle*3 : Tri-n-butyl borane*4 : Poly-méthacrylate de méthyleimmediately.
We recommend ultrasonic cleaning, but if you use a cleaning solution, use aneutral-ph detergent or solution. Use at an appropriate concentration. Use inaccordance with the instruction manual for the cleaning solution. After washing,rinse the solution under sufficient running water and dry it immediately to prevent
corrosion. After cleaning, steam sterilization is performed at 121°C for 15 minutes.
Sterilization should be carried out in accordance with the manufacturer's
instructions for use of the sterilizer. Make sure there is no damage, cracks,
scratches, discoloration, or other abnormalities after cleaning and sterilization.
3 Clean the used Dispensing Stand with commonly used agents and disinfect themin accordance with the commonly used hygienic regulations.

Dental Adhesive Resin Cement

Ciment Dentaire Adhésif à Base de Résine

4.5 Hygieneeach use.

IMPORTANT :READ ALL INSTRUCTIONS THOROUGHLY BEFORE USE.
KEEP THIS LEAFLET AND REFER TO IT PERIODICALLY.

INSTRUCTIONSIMPORTANT :LIRE ATTENTIVEMENT LE MODE D’EMPLOI AVANT L’USAGE.
CONSERVER CETTE BROCHURE ET S’Y RÉFÉRER RÉGULIÈREMENT.

A L’USAGE EXCLUSIF DES DENTISTES

2 After using a Brush Handle, wash it properly and remove any adhering material

INSTRUCTIONS

7) Autres
S'il y a un autre problème, veuillez contacter votre distributeur.

7) Others
If there is any inconvenience, please contact to your distributor.
1 Brush Tips and Dispensing Cups are single use only. Do not reuse. Discard afterENFR

5. Storage

1) Storage conditions
1 Store Super-Bond Brush-dip Starter Kit components at room temperature
between 1 - 30°C and in a dark location. High temperature, high humidity anddirect sunlight will shorten their shelf-life.
2 Do not use this product after the expiration date. See outer package for
expiration date.
2) Product lifetime
Product lifetime can be shortened depending on the circumstances of use. Readall provided instructions carefully before use.

6. Disposal
1 Remove the syringe cap and soak the empty syringe of Super-Bond Brush-dip

Initiator in water 24 hours or longer. Dispose it in accordance with local
regulations.
2 Dispose of empty containers or package except Super-Bond Brush-dip Initiator inaccordance with local regulations.

As in any dental treatments, the patient’s individual constitution and theunique requirements of clinical case at hand must be considered before
selecting materials and conditions for use.
7. How to use
7.1 Surface preparation
It is essential that all surfaces to be bonded with Super-Bond Brush-dip Starter Kit
are properly prepared. Preparation method can vary depending on the nature of thematerials.
1) Tooth surface
1 Cleaning
Clean all surfaces to be bonded following the usual procedure. If necessary,clean all surfaces using a polishing cup / brush with oil-free, fluoride-freepumice. Rinse thoroughly and dry. Isolation by rubber dam or cotton roll isrecommended.
2 Application of Enamel Etchant Gel

1) Remove the syringe cap (red) by twisting it counter-clockwise. Save thesyringe cap.
2) Install the Needle Tip into the syringe certainly and make sure the Needle
Tip and the syringe are connected tightly.
3) Apply the product directly to the tooth surface.
Note: Refer to Table 1 for the application time.
Slowly apply pressure to the syringe plunger. Please push the plunger
gently. The gel may splatter around when strong and fast pressure isapplied.
4) Dispose Needle Tip properly after use and seal the container immediately
with the syringe cap to prevent drying out of the material. Do not store thesyringe with Needle Tip in place. Store it only with the syringe cap.
Note: When applying the Gels directly to the tooth surface, cover the whole
syringe with a disposable barrier (e.g. a plastic film) to prevent
contamination from saliva or blood.
Disinfect the syringe by wiping it with an absorbent gauze padsaturated with disinfectant both before covering and after use.
Table1 Application time
EnamelIndications:Super-Bond Brush-dip Starter Kit est indiqué pour:a. Usage Orthodontiqueb. Fixation directe des dents mobilesc. Bridge collé directementd. Réparation de prothèses cassées

Effets secondaires:
Dans de rares cas, les composants du Super-Bond Brush-dip Starter Kit peuvent
entrainer une sensibilisation. Dans ce cas ne plus utiliser le Super-Bond Brush-dip
Starter Kit.
2. ComposantsSuper-Bond Brush-dip Starter Kit*5 -------------------------- Class IIa[Class IIa]
Composants

Enamel Etchant Gel
30 sec2 Application of primer

Prime the prosthesis if indicated by the manufacturer’s instruction. We
recommend dental prosthesis priming agent manufactured by Sun Medical
Co., Ltd.

7.2 Brush-dip technique
Recommended clinical cases:
Direct fixation of mobile teeth, Bonding orthodontics brackets, Direct bonded
bridge, Repair of fractured prostheses etc.
Note: For various surface preparations, please refer to “7.1 Surface preparation”
above.
1) Dispensing Super-Bond Polymer (Brush-dip Clear)Set two Dispensing Cups to Dispensing Stand. Dispense an appropriate amountof Super-Bond Polymer (Brush-dip Clear) into one of the first cup. Flatten the topof dispensed Super-Bond Polymer (Brush-dip Clear) in the cup by gently tapping
the side of Dispensing Stand with the fingers or hitting the table by the bottom of
the Dispensing Stand.
2) Preparation of Activated Liquid
Hold Super-Bond Brush-dip Monomer bottle perpendicular to the cup anddispense appropriate number of drops into the second cup. Hold Super-Bond
Brush-dip Initiator syringe perpendicular to the cup, dispense the proper numberof drops into the monomer by turning the screw. Stir gently with the Brush Tip.
This mixture is called “Activated Liquid”. The Activated Liquid graduallydeactivates, use it within 5 minutes after mixing.
[Activated Liquid mixing ratio]
Activated Liquid

Super-Bond Brush-dip Monomer

Super-Bond Brush-dip Initiator

8 drops

2 drops

1 drop3) Application of the Super-Bond Brush-dip Starter Kit (resin) cement
Attach Brush Tip to Brush Handle. Wet the Brush Tip with the Activated Liquidthen touch the Super-Bond Polymer (Brush-dip Clear) in the cup lightly with thevery tip of the brush. A small ball of Super-Bond Polymer (Brush-dip Clear) will
form at the end of the brush and can be picked up. Transfer and apply thePolymer ball onto the surface to be bonded. As soon as it touches the surface,
the Polymer will spread out to form a creamy, homogeneous cement layer. If
necessary, repeat the procedure until the entire surface is covered with theadequate amount of cement.
Note: When you repeated the picking up procedure, clean up the brush with gauze
between forming balls.
Note: Apply a coat of the activated liquid to the surface before placing the first layer
will help Super-Bond Brush-dip Starter Kit spread smoothly.
4) Setting and finishing
The curing time varies with the ambient temperature. The standard curing time atmouth temperature (37°C) is 5 to 6 minutes. Start finishing and polishing after 6
minutes or later. Confirm if the sufficient curing has achieved.

Quantité dans chaque coffret
1 × 0.3 mL

1 Super-Bond Brush-dip Initiator

1 × 3.5 mL

2 Super-Bond Brush-dip Monomer

3 Super-Bond Polymer (Brush-dip Clear)1 × 1.5 g

4 Enamel Etchant Gel[Class I

1 × 1 mL]
Accessoires

Quantité dans chaque coffret

5 Dispensing Stand

1

6 Dispensing Cups

20

8 Brush Tips (Brush-dip L)

20

7 Brush Handle (Bent)

1
5

9 23G Needle Tip

Documents inclus : Mode d’emploi, Mode d’emploi illustré, Precautions*5Les composants du coffret sont disponibles séparément
3. Compositions
1 Super-Bond Brush-dip Initiator:
Tri-n-butyle borane partiellement oxidé, Autres.
2 Super-Bond Brush-dip Monomer:
Monomères de Méthacrylate, Autres.
3 Super-Bond Polymer (Brush-dip Clear):
Poly méthyle méthacrylate, Autres.
4 Enamel Etchant Gel:
Acide phosphorique , eau, épaississant, pigment, Autres.
Composants
1
2

Forme

Liquide transparent incolore

Liquide transparent incolore jaune pâle3

après utilisation.

2 Après utilisation d’un manche de brosse, lavez-le et retirez correctement toutematière adhérente.
Nous recommandons un nettoyage par ultrasons, mais en cas d’utilisation d’unesolution de nettoyage, utilisez un détergent ou une solution à pH neutre. Utilisez
à concentration adéquate. Utilisez conformément au mode d’emploi de la solution
de nettoyage. Après nettoyage, rincez la solution avec suffisamment d’eau etséchez-la immédiatement pour éviter la corrosion.
Après nettoyage, une stérilisation à la vapeur est effectuée à 121°C durant 15
minutes. La stérilisation doit être effectuée conformément au mode d’emploi dufabricant du stérilisateur.
Assurez-vous qu’il n’y a aucune présence de dommages, de fissures, de rayures,de décoloration ou autre anomalie après nettoyage et stérilisation.
3 Nettoyez le Dispensing Stand avec des agents à usage courant et désinfectez-lesconformément aux règles d’hygiène courantes.
5. Stockage1) Conditions de stockage
1 Stocker les composants Super-Bond Brush-dip Starter Kit à températureambiante entre 1 et 30 ° C et dans un endroit sombre. Une température élevée,
une humidité élevée et la lumière directe du soleil réduiront leur durée de vie.
2 Ne pas utiliser ce produit après la date de péremption. Voir l'emballage pour ladate de péremption.
2) Durée de vie du produit
La durée de vie du produit peut être raccourcie en fonction des circonstancesd'utilisation. Lisez attentivement toutes les instructions fournies avant utilisation.

Contre-indications:Patient présentant des réactions allergiques sévères à ce produit ou aux
monomères de méthacrylate et à l’acétone.

42) Metal, ceramic, zirconia and resin composite surface
1 Cleaning and conditioning
Clean all surfaces to be bonded following the usual procedure. Pretreat andcondition the surface to be bonded, following the specific instructions of each
prosthetic material and product. Roughen up the surface with diamond bur or
sandblast if necessary.
Note: Applying Enamel Etchant Gel will help remove surface contaminants,
and ultrasonic cleaner is also effective.

4 drops

Usage prévu:
Super-Bond Brush-dip Starter Kit est conçu pour adhérer à l’émail dentaire, lemétal, la dentine, la céramique, la zircone et la résine composite avec lepré-traitement approprié.

4.5 Hygiène

1 Les Brush Tips et Dispensing Cups sont à usage unique. Ne pas réutiliser. Jeter

Poudre blanche
Gel rouge

4. Precautions

4.1 Pour la sécurité

1 Les composants du Super-Bond Brush-dip Starter Kit contiennent des substanc-es, telles que les monomères de méthacrylate, qui peuvent causer des réactions
allergiques. Ils ne doivent pas être utilisés sur ou par des personnes sensibles
aux acrylates, méthacrylates ou matériaux similaires. L’exposition à des résines
non polymérisées peut causer des irritations, dermatites ou des réactions
inflammatoires des tissus (peau, muqueuse, yeux etc.) chez les individus
sensibles. Si cela arrive, stopper l’usage, rincer la zone avec beaucoup d’eau etconsulter un médecin si nécessaire.
2 Eviter les contacts des composants Super-Bond Brush-dip Starter Kit avec lestissus mous, la peau et les yeux. La zone contaminée doit être immédiatement
essuyée avec de l'alcool, puis rincée abondamment à l'eau courante. Si lescomposants touchent l‘œil, rincer immédiatement et abondamment à l'eau
courante et faire examiner le patient par un ophtalmologiste.
3 Évitez toute ingestion accidentelle lors de l'application et du rinçage descomposants Super-Bond Brush-dip Starter Kit. En cas d'ingestion, consulter unmédecin, si nécessaire.
4 Manipulez les composants Super-Bond Brush-dip Starter Kit dans un endroit bien
ventilé (ventiler la zone plusieurs fois par heure). En cas d'inhalation, sortir à l'air frais.
5 Les Super-Bond Brush-dip Initiator et du Super-Bond Brush-dip Monomer sonttrès volatils et inflammables. Ne pas les stocker dans un endroit où ils peuventêtre exposés à une flamme libre.
4.2 Précautions communes

1 L‘usage d’une protection en caoutchouc est recommandé.
2 Le dentiste doit porter des gants de protection.
3 Ne pas mélanger les composants avec d’autres matériaux.
4 Utiliser ce produit seulement pour les indications recommandées.

4.3 Précautions de manipulation
1 Une surface contaminée par de l ‘huile, du sang, de la salive ou des biofilms
diminue la force d’adhésion. Bien nettoyer la dent et les prothèses. Ensuite,
prendre garde à ne pas recontaminer en isolant la surface correctement. L ‘usage
d’une protection en caoutchouc est recommandée.
2 Les fluorures et l‘huile diminuent la force d’adhésion. Une pâte dentaire contenant del‘huile ou du fluor ne doit pas être utilisée pour le nettoyage de la surface de la dent.
3 L’Eugénol est un inhibiteur de polymérisation. Des bases et ciments contenant del‘eugénol ne doivent pas être utilisés avec les composants du Super-Bond
Brush-dip Starter Kit.
4 L’Hypochlorite de Sodium est un inhibiteur de polymérisation. Quand vous utilisez
Super-Bond Brush-dip Starter Kit sur la surface dentine traitée avec del’Hypochlorite de Sodium, le Super-Bond C&B Green Activator ou le Dentin
Etchant Gel doivent être appliqués après l’usage de l’hypochlorite de Sodium. La
durée du traitement avec l’Hypochlorite de Sodium ne doit pas dépasser 15
secondes. Les détails sur l ‘utilisation des Super-Bond C&B Green Activator et
Dentin Etchant Gel sont inclus dans le document fourni.
5 Les Super-Bond Brush-dip Initiator et du Super-Bond Brush-dip Monomer sonttrès volatils. Bien refermer les produits immédiatement après usage.
6 Le liquide activé (mélange du Super-Bond Brush-dip Initiator et du Super-Bond
Brush-dip Monomer) perd ses propriétés dans le temps et doit être utilisé dans les
5 minutes après le mélange.
7 Super-Bond Brush-dip Starter Kit peut être décoloré s'il est associé à un cimenttemporaire contenant de l'acide tannique.
8 Les composants Super-Bond Brush-dip Starter Kit sont uniquement à usage
unique. Ne pas remettre les composants utilisés dans leur emballage d'origine.
Éliminer les restes de matériau soigneusement. Après utilisation de Super-Bond
Brush-dip Starter Kit avec le pinceau à tremper, jetez l'excès de Super-Bond
Polymer (Brush-dip Clear) restant dans le godet.
9 Les Brush Tips et Dispensing Cups sont à usage unique. Ne pas les réutiliser. A
jeter après chaque usage.
10 Quand une préparation est en profondeur, proche de la pulpe, protéger la pulpe
avec un matériau approprié.
11 Avant d'utiliser les accessoires, vérifiez la présence de dommages, de fissures,de rayures ou de décoloration, et cessez de les utiliser en cas d'anomalie.
4.4 Précautions pour le Super-Bond Brush-dip Initiator
Exposé à l ‘air ou à l ‘eau, le Super-Bond Brush-dip Initiator perd de l’activité et peut
générer de la chaleur. Respectez les précautions suivantes.
1) Conditions de stockage
Évitez les températures élevées, l'humidité et la lumière directe du soleil. Le
changement répété de température peut entraîner l’aspiration d’air dans laseringue, réduisant ainsi la durée de vie du Super-Bond Brush-dip Initiator.
*Après un stockage prolongé, la première goutte du Super-Bond Brush-dip
Initiator peut avoir une activité moindre en raison de l'exposition à l'air (oxygène),bien que le reste du matériau contenu dans le récipient ne soit pas affecté.
L'élimination d'une goutte avant l’utilisation assurera la réaction chimiqueappropriée.
*La seringue est en verre. Par conséquent, elle doit être manipulée avec soinpour éviter les chocs, les chutes et autres dommages physiques.
2) Après l ‘usage
Tournez la vis-piston de la seringue deux fois dans le sens anti-horaire après
chaque utilisation pour relâcher la pression dans le cylindre du Super-Bond
Brush-dip Initiator. (La pression accumulée peut provoquer une fuite duSuper-Bond Brush-dip Initiator ou, dans certains cas, fissurer la seringue).
3) Fermeture du capuchon de la seringue
Le capuchon glisse tout simplement. Recapuchonner la seringue immédiatement
après chaque utilisation. L'air désactive le Super-Bond Brush-dip Initiator. Ne
jamais laisser la seringue SANS le capuchon.
4) Nettoyage de la seringue
Essuyez le bout de la seringue avec une compresse sèche après chaqueutilisation pour éviter l’accumulation de résidus. La compresse utilisée doit êtresaturée d’eau pour désactiver complètement le Super-Bond Brush-dip Initiator
avant de la jeter. L'accumulation de résidus peut empêcher le repositionnement
correct du capuchon de fermeture.
5) Nettoyer le Super-Bond Brush-dip Initiator renversé immédiatement avec
des serviettes humides
Si le Super-Bond Brush-dip Initiator est absorbé par un matériau combustible, ilpeut augmenter suffisamment la température pour provoquer une combustion
lente. Le Super-Bond Brush-dip Initiator renversé ou non utilisé doit être immédiatement tamponné avec une serviette jetable HUMIDE (pas sèche). Ensuite,
saturer la serviette jetable à l’eau courante pour désactiver complètement leSuper-Bond Brush-dip Initiator à l’eau courante.
6) Problème pour exprimer le Super-Bond Brush-dip Initiator
Si le Super-Bond Brush-dip Initiator ne forme pas de goutte en raison du blocage
de l'orifice ou si la vis ne tourne pas, ne pas forcer. Le contenu peut éclabousser
si la seringue en verre se casse. Veuillez contacter votre distributeur.

6. Elimination
1 Retirer le capuchon de la seringue et faire tremper la seringue vide deSuper-Bond Brush-dip Initiator dans de l'eau pendant 24 heures ou plus. Éliminer
conformément à la réglementation locale.
2 Éliminer les récipients ou les emballages vides, à l'exception du Super-Bond
Brush-dip Initiator, conformément à la réglementation locale.

Comme dans tout traitement dentaire, la constitution individuelle du patient
et les exigences spécifiques du cas clinique doivent être considérés avantde sélectionner les matériaux et les conditions d'utilisation.
7. Comment l’utiliser
7.1 Préparation de la surface
Il est essentiel de bien préparer toutes les surfaces devant être collées au
Super-Bond Brush-dip Starter Kit. La méthode de préparation peut varier selon lanature du support.
1) Surface de la dent
1 Nettoyage
Nettoyer toutes les surfaces à coller selon la technique habituelle. Si
nécessaire, nettoyez toutes les surfaces à l’aide d’une cupule de polissage
avec de la pierre ponce sans huile ni fluor. Rincez soigneusement et séchez.
L'isolement par une protection en caoutchouc ou des rouleaux de coton est
recommandé.
2 Application du Enamel Etchant Gel

1) Retirer le bouchon de la seringue (rouge) en le dévissant. Conserver cebouchon.
2) Installez le Needle Tip dans la seringue et assurez-vous que le Needle Tip
et la seringue sont fermement connectés.
3) Appliquer le produit sur la surface de la dent.
Note: Voir le tableau 1 pour les temps d’application.
Appliquer une pression douce sur le piston de la seringue. Toujours
pousser doucement sur le piston. Le gel peut éclabousser lorsqu’unepression forte et rapide est appliquée.
4) Jetez le Needle Tip après usage et fermez immédiatement le recipient avec
le capuchon de seringue pour éviter que le contenu ne sèche. Ne stockez
pas la seringue avec le Needle Tip connecté. Stockez-la uniquement avec lecapuchon de seringue.
Note: Lors de l ‘application des Gels directement sur la surface de la dent
,protéger la seringue avec une protection jetable (film plastique par
exemple) afin de prévenir toute contamination par la salive ou le sang.
Désinfecter la seringue en l ‘essuyant avec une compresse saturée dedésinfectant avant de la protéger et après l ‘usage.
Table1 Temps d’application
Email

Enamel Etchant Gel
30 secGEBRAUCHSANLEITUNGWICHTIG:ANLEITUNG VOR GEBRAUCH SORGFÄLTIG LESEN.
DIE ANLEITUNG AUFHEBEN UND REGELMÄSSIG ZU RATE ZIEHEN.

Adhäsivresinzement

Super-Bond Brush-dip Starter Kit

2 Application du primer

Appliquer un primer sur la prothèse si le instructions du fabricant lerecommande. Nous recommandons l'agent d'amorçage de prothèse dentaire
fabriqué par Sun Medical Co., Ltd.

7.2 Technique du pinceau
Cas cliniques recommandés :Fixation directe de dents qui bougent, collage de brackets orthodontiques, bridges
collés, réparation de prothèses cassées, etc.
Note: Pour la préparation de surfaces variées se referrer au chapitre « 7.1
préparation des surfaces » ci-dessus.
1) Utilisation de la Super-Bond Polymer (Brush-dip Clear)
Insérer deux Dispensing Cups dans le Dispending Stand. Déposer la quantité
appropriée de Super-Bond Polymer (Brush-dip Clear) dans la première coupelle.
Aplanir la surface de la poudre en tapotant avec le doigt le côté du DispensingStand ou bien tapoter le dessous du Dispensing Stand sur la table.
2) Préparation du liquide activé
Tenir le flacon de Super-Bond Brush-dip Monomer perpendiculairement à lacoupelle et déposer le nombre de gouttes souhaité dans la deuxième coupelle.
Tenir la seringue de Super-Bond Brush-dip Initiator perpendiculairement à lacoupelle, et déposer le nombre de gouttes souhaité dans le monomère en
vissant le piston-vis. Remuez doucement avec la pointe du Brush Tip. Cemélange s'appelle « Liquide Activé ». Le Liquide Activé se désactive
progressivement, utilisez-le dans les 5 minutes qui suivent le mélange.
[Rapport de mélange du Liquide Activé]
Liquide Activé

Super-Bond Brush-dip Monomer

Super-Bond Brush-dip Initiator

8 gouttes

2 gouttes

1 goutte

3) Application du ciment résine Super-Bond Brush-dip Starter Kit
Placer un Brush Tip sur le Brush Handle. Tremper le Brush Tip dans le Liquide
Activé puis touchez la Super-Bond Polymer (Brush-dip Clear) dans la coupelle
avec le bout de la brosse. Une petite boule de Super-Bond Polymer (Brush-dip
Clear) se formera au bout de la brosse et peut être recueillie. Prenez-là etappliquez la boule de Polymère sur la surface à coller. Dès que la boule touche
la surface, le Polymère s’étale pour donner une couche de ciment crémeuse ethomogène. Si nécessaire répéter la procédure jusqu’à recouvrir toute la surface
à coller de la quantité suffisante de ciment.
Note: Quand la procédure est répétée, il faut impérativement nettoyer le Brush Tip
avec une compresse avant de reformer une nouvelle boule de résine.
Note: Appliquer une couche de liquide activé sur la surface avant la première
couche aidera à étaler en douceur le Super-Bond Brush-dip Starter Kit.
4) Temps de prise et finition
Le temps de prise varie selon la température ambiante. Le temps de prisestandard à la température buccale (37°C) est de 5 à 6 minutes. Démarrer lafinition et le polissage après 6 mn ou plus tard. Vérifiez que la prise suffisante a
été obtenue.

7) Sonstiges
Bei Problemen bitte den Händler kontaktieren.
4.5 Hygiene

1 Brush Tips und Dispensing Cups sind nur für den Einmalgebrauch. Nicht wieder
2 Den Pinselhalter nach der Verwendung gründlich waschen. Entfernen Sie

1. Produktüberblick
Super-Bond Brush-dip Starter Kit ist ein selbsthärtender adhäsiver Resinzement,
der 4-META*1, MMA*2, TBB*3 und PMMA*4 (4-META / MMA-TBB-Resin) enthält.
Super-Bond Brush-dip Starter Kit kann für die Befestigung aller festsitzenden
Zahnprothesen aus unterschiedlichsten dentalen Materialien verwendet werden.
*14-Metacryloyloxyethyl-trimellitate Anhydrid*2Methylmethacrylat*3Tri-n-butylboran*4Polymethylmethacrylat
Verwendungszweck:
Super-Bond Brush-dip Starter Kit ist für das Kleben auf Zahnschmelz, Metall,
Keramik, Zirkonoxid und Resinkomposit vorgesehen, vorausgesetzt die jeweilige
Oberfläche wurde entsprechend vorbehandelt.
Indikationen:
Super-Bond Brush-dip Starter Kit ist für folgende Anwendungen indiziert:
a. Kieferorthopädische Befestigungenb. Direkte Fixierung gelockerter Zähne
c. Klebebrücken
d. Reparatur gebrochener Zahnprothesen

Kontraindikation:
Patienten mit bekannten schweren allergischen Reaktionen gegenüber diesem
Produkt oder Methacrylat-Monomeren und Azetonen.
Nebenwirkungen:
In seltenen Fällen können Komponenten von Super-Bond Brush-dip Starter Kit zu
einer Allergisierung führen. In solchen Fällen sollte das Produkt nicht mehr
angewendet werden.

anhaftendes Material sofort.
Wir empfehlen eine Ultraschallreinigung, aber falls Sie eine Reinigungslösung
verwenden, sollten Sie eine Lösung oder ein Reinigungsmittel verwenden, das
pH-neutral ist. Benutzen Sie das Mittel in einer angemessenen Konzentration.
Benutzen Sie es gemäß der Gebrauchsanweisung für diese Reinigungslösung.
Spülen Sie nach dem Waschen die Lösung mit ausreichend fließendem Wasser
aus und trocknen Sie den Pinselhalter sofort, um Korrosion zu vermeiden.
Führen Sie nach der Reinigung für 15 Minuten eine Dampfsterilisation bei 121°C
durch. Diese Sterilisation sollte gemäß den Vorgaben des Herstellers bezüglich
der Verwendung des Sterilisators durchgeführt werden.
Stellen Sie sicher, dass nach der Reinigung und Sterilisation keine Beschädigungen, Risse, Kratzer, Verfärbungen oder andere Unregelmäßigkeit vorliegen.
3 Reinigen Sie Dispensing Stand mit den üblichen Mitteln und desinfizieren Sie sie
gemäß den üblichen Hygiene-Vorschriften.
5. Lagerung

1) Lagerbedingungen1 Super-Bond Brush-dip Starter Kit-Komponenten im Dunkeln bei einer Raumtemperatur zwischen 1-30 °C lagern. Hohe Temperaturen, hohe Luftfeuchtigkeit unddirekte Sonneneinstrahlung verkürzen die Haltbarkeitsdauer.
2 Das Produkt nicht nach Ablauf des Haltbarkeitsdatums verwenden.
Haltbarkeitsdatum: siehe äußere Verpackung.
2) Lebensdauer des Produkts
Die Produktlebensdauer kann sich je nach den Umständen des Gebrauchs
verkürzen. Vor Gebrauch sämtliche Anweisungen gründlich lesen.

6. Entsorgung
1 Die Spritzenkappe entfernen und die leere Spritze des Super-Bond Brush-dip

2. Komponenten
Super-Bond Brush-dip Starter Kit* ------------------------ Klasse IIa
5[Klasse IIa]
Komponenten

Inhalt pro Kit

1 Super-Bond Brush-dip Initiator

1 × 0,3 mL

2 Super-Bond Brush-dip Monomer

3 Super-Bond Polymer (Brush-dip Clear)1 × 3,5 mL
1 × 1,5 g

4 Enamel Etchant Gel[Klasse I

1 × 1 mL]
Zubehör

Inhalt pro Kit

5 Dispensing Stand

1

6 Dispensing Cups

20

8 Brush Tips (Brush-dip L)

20

7 Brush Handle (Bent)

1
5

9 23G Needle Tip

Beiliegende Dokumente: Anleitung, bebilderte Anleitungskarte, Vorsichtsmaßnahmen*5Die Komponenten des Kits können einzeln nachgekauft werden.
3. Zusammensetzung:1 Super-Bond Brush-dip Initiator:
Teiloxidiertes Tri-n-Butylboran, Sonstiges.
2 Super-Bond Brush-dip Monomer:
Methacrylat-Monomer, Sonstiges.
3 Super-Bond Polymer (Brush-dip Clear):
Polymethyl-Methacrylat, Sonstiges.
4 Enamel Etchant Gel:
Phosphorsäure, Wasser, Verdicker, Pigment, Sonstiges.
Form

1
2

6) Problem beim Ausbringen von Super-Bond Brush-dip Initiator
Wenn der Super-Bond Brush-dip Initiator aufgrund einer Verstopfung der Öffnung
keinen Tropfen bildet oder sich der Schraubverschluss nicht dreht, diesen nicht
mit Gewalt drehen. Bei einem Bruch der Glasspritze kann der Inhalt herausspritzen. Bitte den Händler kontaktieren.

verwenden, sondern nach jedem Gebrauch entsorgen.

NUR FÜR DEN ZAHNÄRZTLICHEN GEBRAUCH

Komponenten

2) Surfaces en métal, céramique, zircone et résine composite
1 Nettoyage et préparation
Nettoyer toutes les surfaces à coller selon la procédure habituelle. Préparez ettraitez la surface à coller selon les instructions propres à chaque matériau deprothèse et chaque produit. Rendre rugueuse la surface avec une fraisediamantée ou sabler si nécessaire.
Note: Appliquer le Enamel Etchant Gel aide à retirer les contaminants desurface et le nettoyage par ultrasons est aussi efficaces.

4 gouttes

DE

farblose durchsichtige Flüssigkeit

Initiator 24 Stunden oder länger in Wasser legen. Anschließend die Spritze
gemäß den vor Ort geltenden Bestimmungen entsorgen.
2 Die leeren Behälter oder Packungen, außer Super-Bond Brush-dip Initiator,
gemäß den vor Ort geltenden Bestimmungen entsorgen.

Wie bei jeder Zahnbehandlung müssen vor Auswahl der Materialien und
Anwendungsbedingungen die individuelle Konstitution des Patienten sowie
die speziellen Gegebenheiten des klinischen Falles berücksichtigt werden.
7. Anwendung
7.1 Oberflächenpräparation
Es ist unabdingbar, sämtliche mit Super-Bond Brush-dip Starter Kit zu
verklebenden Flächen korrekt vorzubehandeln. Die Präparationsmethode kannentsprechend der Beschaffenheit des Materials variieren.
1) Zahnoberfläche
1 Reinigung
Sämtliche zu verklebenden Oberflächen wie üblich reinigen. Falls erforderlich,
die Oberflächen mit einem Polierkelch / -bürstchen und einer öl- undfluoridfreien Paste reinigen. Danach gründlich spülen und trocknen. Eine
Isolierung mittels Kofferdam oder Zahn-Watterolle wird empfohlen.
2 Auftrag von Enamel Etchant Gel1) Öffnen des Spritzenverschlusses durch Drehen gegen den Uhrzeigersinn.
Verschluss nicht verwerfen.
2) Befestigen Sie Needle Tip sachgemäß auf der Spritze und vergewissern Sie
sich, dass Needle Tip fest mit der Spritze verbunden ist.
3) Produkt direkt auf die Zahnoberfläche auftragen
Hinweis: Applikationszeit auf Tabelle 1 beachten.
Langsamen Druck auf den Spritzenkolben ausüben. Bei zu
starkem und zu schnellem Druck kann das Gel verspritzen.
4) Entsorgen Sie Needle Tip nach der Verwendung sorgfältig und verschließen
Sie den Container sofort mit der Spritzenkappe, um ein Austrocknen desMaterials zu verhindern. Bewahren Sie die Spritze nicht mit aufgesetzter
Needle Tip auf. Bewahren Sie sie nur mit Spritzenkappe auf.
Hinweis: im Falle des Auftrag des Gels direkt auf die Zahnoberfläche diegesamte Spritze mit einem Einwegplastikfilm / -hülle bedecken, um
eine Kontamination mit Speichel oder Blut zu vermeiden. Spritze
mit einem weichen in Desinfektionsmittel getränktem Tuch vor undnach Gebrauch abwischen.
Table1 Applikationszeit

farblose bis blassgelbe durchsichtige Flüssigkeit

3
4

weißes Pulverrotes Gel

4. Vorsichtsmaßnahmen:

4.1 Sicherheitsmaßnahmen

1 Super-Bond Brush-dip Starter Kit enthält Substanzen wie Methacrylat-Monomere,

die eine allergische Reaktion hervorrufen können. Sie sind nicht anzuwenden bei
Personen, die allergisch gegenüber Acrylaten, Methacrylaten und ähnlichen
Materialien sind. Der Kontakt mit nichtgehärtetem Resin kann bei empfindlichen
Personen zu Reizungen, Dermatitis oder entzündlichen Reaktionen desGewebes (Haut, Schleimhaut, Augen usw.) führen. Falls dies der Fall ist, den
Gebrauch stoppen, die betroffenen Stellen mit reichlich Wasser spülen und falls
notwendig medizinischen Rat einholen.
2 Den Kontakt der Super-Bond Brush-dip Starter Kit-Komponenten mit Weichgewebe, Haut und Augen vermeiden. Kontaminierte Stellen sofort mit Alkohol
abwischen und anschließend mit reichlich fließendem Wasser spülen. Falls
Komponenten ins Auge gelangen, sofort gründlich unter reichlich fließendem
Wasser spülen und den Patienten einen Augenarzt aufsuchen lassen.
3 Ein versehentliches Verschlucken während des Auftrags und Spülens vonSuper-Bond Brush-dip Starter Kit-Komponenten vermeiden. Kommt es dennoch
dazu, gegebenenfalls medizinischen Rat einholen.
4 Super-Bond Brush-dip Starter Kit-Komponenten in einem gut belüfteten Raum
anwenden (Raum mehrfach pro Stunde lüften). Im Falle des Einatmens derDämpfe an die frische Luft gehen.
5 Super-Bond Brush-dip Initiator und Super-Bond Brush-dip Monomer sind starkflüchtig und leicht entflammbar. Nicht in einem Raum lagern, in dem die Komponenten einer offenen Flamme ausgesetzt sein könnten.
4.2 Allgemeine Vorsichtsmaßnahmen

1 Der Gebrauch eines Kofferdams wird empfohlen.
2 Der Behandler sollte Handschuhe tragen.
3 Die Komponenten nicht mit denen anderer Materialien mischen.
4 Das Produkt nur für den in der Anleitung empfohlenen Gebrauch anwenden.

4.3 Vorsichtsmaßnahmen während der Anwendung

1 Oberflächen, die mit Öl, Blut, Speichel und Biofilm kontaminiert sind, verringern

die Haftkraft. Zahn und Zahnprothese gründlich reinigen, danach Rekontamination durch Isolierung der Oberfläche vermeiden. Der Gebrauch eines Kofferdams
wird sehr empfohlen.
2 Fluorid und ölige Substanzen verringern die Haftkraft. Pasten, die Fluorid oder Öl
enthalten, sollten nicht zur Reinigung der Oberflächen benutzt werden.
3 Eugenol ist ein Polymeriationshemmer. Aus diesem Grund eugenolhaltige
Untergründe und Zemente nicht zusammen mit den Super-Bond Brush-dip
Starter Kit-Komponenten verwenden.
4 Natriumhypochlorit ist ein Polymerisationshemmer. Falls Super-Bond Brush-dip
Starter Kit auf einer mit Natriumhypochlorit behandelten Dentinoberfläche
angewendet wird, sollte nach dem Gebrauch von Natriumhypochlorit derSuper-Bond C&B Green Activator oder das Dentin Etchant Gel aufgetragenwerden. Die Behandlungsdauer mit Natriumhypochlorit sollte weniger als 15
Sekunden betragen. Angaben, wie der Super-Bond C&B Green Activator und das
Dentin Etchant Gel angewendet werden, finden sich im beiliegenden Dokument.
5 Super-Bond Brush-dip Initiator und Super-Bond Brush-dip Monomer sind leicht
flüchtig. Daher nach Gebrauch sofort verschließen.
6 Die aktivierte Flüssigkeit (Mischung aus Super-Bond Brush-dip Initiator undSuper-Bond Brush-dip Monomer) deaktiviert sich nach und nach. Daher innerhalb
von 5 Minuten nach Anmischen gebrauchen.
7 Super-Bond Brush-dip Starter Kit kann sich verfärben, wenn es mit einem
temporären Zement kombiniert wird, der Tanninsäure enthält.
8 Ausgebrachte Komponenten nicht wieder in den Originalbehälter einbringen.
Übrig gebliebenes Material ordnungsgemäß entsorgen. Bei Anwendung vonSuper-Bond Brush-dip Starter Kit mit der Brush-Dip-Technik in der Schale
verbliebenes Super-Bond Polymer (Brush-dip Clear) entsorgen.
9 Brush Tips und Dispensing Cups sind nur für den Einmalgebrauch vorgesehen.
Nicht wieder verwenden, sondern nach jedem Gebrauch entsorgen.
10 Bei einer tiefen Präparation bis nahe an die Pulpa die Pulpa mit geeignetem
Material schützen.
11 Prüfen Sie vor der Verwendung das Zubehör auf Beschädigungen, Risse,
Kratzer, Verfärbungen etc. und verwenden Sie es nicht, falls Sie irgendeine
Unregelmäßigkeit feststellen.
4.4 Vorsichtsmaßnahme bei Super-Bond Brush-dip Initiator
Wird der Super-Bond Brush-dip Initiator Luft oder Wasser ausgesetzt, kann er an
Aktivität verlieren und Hitze erzeugen. Daher sind die nachfolgend aufgeführten
Regeln zu beachten.
1) Lagerbedingungen
Hohe Temperaturen, hohe Luftfeuchtigkeit und direkte Sonneneinstrahlung sindzu vermeiden. Wiederholte Temperaturwechsel können bei der Spritze zu
Luftansaugung führen, wodurch sich die Haltbarkeitsdauer des Super-Bond
Brush-dip Initiator verkürzt.
*Nach längerer Lagerung kann der erste Tropfen des Super-Bond Brush-dip
Initiator aufgrund der Exposition in Luft (Sauerstoff) weniger Aktivität aufweisen;der Rest des Behälters ist davon nicht betroffen. Für eine korrekte chemische
Reaktion empfiehlt sich daher die Entsorgung des ersten Tropfens.
*Da die Spritze aus Glas ist, muss sie sorgfältig behandelt werden, um Stöße, ein
Herunterfallen oder andere physische Beschädigungen zu vermeiden.
2) Nach Gebrauch
Nach jedem Gebrauch die Spritzenschraube zwei Mal gegen den Uhrzeigersinn
drehen, um den Druck im Zylinder der Super-Bond Brush-dip Initiator-Spritze
abzubauen. (Ein Druckaufbau in der Spritze kann zum Entweichen vonSuper-Bond Brush-dip Initiator oder in manchen Fällen zu Sprüngen in derSpritze führen.)3) Verschlusskappe
Die Verschlusskappe lässt sich leicht aufsetzen und abziehen. Die Kappe
unbedingt nach jedem Gebrauch auf die Spritze setzen. Luft deaktiviert
Super-Bond Brush-dip Initiator.
4) Reinigen der Spritze
Nach jedem Gebrauch die Spitze der Spritze mit trockener Gaze abwischen, um
die Ansammlung von Restmaterial zu vermeiden. Die Gaze ist vorher mit Wasser
zu durchtränken, um den Super-Bond Brush-dip Initiator vor seiner Entsorgung zu
deaktivieren. Wenn sich an der Spitze der Spritze zu viel Restmaterial ansammelt, führt dies zu einem unsachgemäßen Sitz der Verschlusskappe.
5) Verschütteten Super-Bond Brush-dip Initiator unverzüglich mit nassem
Tuch aufnehmen
Wird der Super-Bond Brush-dip Initiator von einem entflammbaren Material
aufgenommen, kann sich die Temperatur so weit erhöhen, dass es zu einem
Schwelbrand kommt. Verschütteter oder nicht gebrauchter Super-Bond Brush-dip
Initiator sollte unverzüglich mit einem NASSEN (nicht trockenen) Einwegtuch
abgetupft werden. Danach das Tuch unter laufendes Wasser halten, um den
Super-Bond Brush-dip Initiator zu deaktivieren.

Schmelz

Enamel Etchant Gel
30 s2) Metall-, Keramik,- Zirkonoxid- und Resinkomposit-Oberflächen
1 Reinigung und Konditionierung
Sämtliche zu verklebenden Oberflächen wie üblich reinigen. Vorbehandlung
und Konditionierung der zu verklebenden Oberfläche gemäß den speziellen
Anweisungen für das jeweilige Produkt. Die Oberfläche mit einem
Diamantbohrer aufrauen oder falls erforderlich sandstrahlen.
Hinweis: Die Anwendung von Enamel Etchant Gell hilft bei der Entfernung vonOberflächenverunreinigungen; eine Ultraschallreinigung ist ebenfalls
wirksam.
2 Auftrag von Primer

Grundieren Sie die Prothese, wenn dies in der Anleitung des Herstellers
entsprechend angewiesen wird. Wir empfehlen die Verwendung eines
Verbindungssystems von Sun Medical Co., Ltd..

7.2 Brush-Dip-Technik
Empfohlene klinische Anwendungen:Direkte Fixierung von lockeren Zähnen, Kleben von kieferorthopädischen Brackets,
Klebebrücken, Reparatur von gebrochenen Prothesen usw.
Hinweis: In Bezug auf die unterschiedlichen Oberflächenpräparationen bitte
Abschnitt 7.1 „Oberflächenpräparation“ beachten.
1) Ausbringen des Super-Bond Polymer (Brush-dip Clear)
Zwei Dispensing Cups in den Dispensing Stand stellen. Die erforderliche Menge
Super-Bond Polymer (Brush-dip Clear) in den ersten Dosierbecher ausbringen.
Um die Oberfläche des Pulvers zu glätten, mit den Fingern leicht an den Rand
des Dispensing Stand klopfen oder mit der Hand auf den Tisch, auf dem derDispensing Stand steht, klopfen.
2) Vorbereitung der aktivierten Flüssigkeit
Die Super-Bond Brush-dip Monomer Flasche senkrecht zum zweiten
Dosierbecher halten und die erforderliche Menge an Tropfen in den
Dosierbecher ausbringen. Die Super-Bond Brush-dip Initiator Spritze senkrecht
zum Dosierbecher halten, die korrekte Anzahl an Tropfen durch Drehen desSchraubverschlusses in das Monomer einbringen. Vorsichtig mit dem Brush Tip
verrühren. Diese Mischung wird als „aktivierte Flüssigkeit“ bezeichnet. Die
aktivierte Flüssigkeit deaktiviert sich allmählich, daher sollte die Anwendung
innerhalb von 5 Minuten nach dem Anmischen erfolgen.
[Mischverhältnis für die aktivierte Flüssigkeit]
Aktivierte Flüssigkeit

Super-Bond Brush-dip Monomer

Super-Bond Brush-dip Initiator

8 Tropfen

2 Tropfen

4 Tropfen

1 Tropfen

3) Auftrag von Super-Bond Brush-dip Starter Kit-Resinzement
Den Brush Tip am Brush Handle befestigen. Den Brush Tip mit der aktivierten
Flüssigkeit befeuchten, danach mit der Spitze des Pinsels das Super-Bond
Polymer (Brush-dip Clear) leicht berühren. An der Spitze des Pinsels bildet sichein kleines Kügelchen des Pulvers, das aufgenommen werden kann. Diese
Polymer-Kugel auf die zu verklebende Oberfläche auftragen. Sobald das
Kügelchen die Oberfläche berührt, breitet sich das Polymer zu einer cremigenhomogenen Zementschicht aus. Falls erforderlich, den Vorgang wiederholen, bis
die gesamte Oberfläche mit einer ausreichenden Menge an Zement bedeckt ist.
Hinweis: Bei wiederholter Anwendung den Pinsel jeweils mit Gaze säubern.
Hinweis: Vor dem Auftragen der ersten Schicht Zement kann es hilfreich sein, eine
Schicht der aktivierten Flüssigkeit aufzutragen. Dadurch kann sich derSuper-Bond Brush-dip Starter Kit besser verteilen.
4) Aushärtung und Fertigstellung
Die Aushärtungszeit ist von der Umgebungstemperatur abhängig. Die
Standardaushärtungszeit bei Mundtemperatur (37°C) beträgt 5-6 Minuten. Mit
der Endbearbeitung und dem Polieren darf frühestens nach 6 Minuten begonnen
werden. Eine Prüfung, ob die Aushärtung ausreichend ist, wird angeraten.

I T

4.5 Misure igieniche

1 Le Brush Tip e le Dispensing Cup sono monouso. Non riutilizzarle. Eliminarle

dopo l’uso.

ISTRUZIONIIMPORTANTE:LEGGERE ATTENTAMENTE LE ISTRUZIONI PRIMA DELL'USO.
CONSERVARE QUESTO FOGLIETTO E FARE RIFERIMENTO AD ESSO PERIODICAMENTE.

Cemento Dentale a Base di Resina Adesiva

Super-Bond Brush-dip Starter Kit
SOLO PER USO DENTISTICO
1. Presentazione del prodottoSuper-Bond Brush-dip Starter Kit è un cemento a base di resina adesiva
autopolimerizzante, contiene 4-META*1, MMA*2, TBB*3 e PMMA*4 (resina 4-META /MMA-TBB). Può essere utilizzato per cementare tutte le protesi dentali fisse
fabbricate con una vasta gamma di materiali dentali.
*14-metacriloilossietil trimellitato anidride*2Metil metacrilato*3Tri-n-butilborano*4Poli metil metacrilato
Uso previsto:
Super-Bond Brush-dip Starter Kit aderisce allo smalto dentale, metallo, ceramica,
zirconio e resina composita dopo un appropriato pretrattamento della superficie.
Indicazioni:
Super-Bond Brush-dip Starter Kit è indicato per:a. Applicazioni ortodonticheb. Fissaggio diretto dei denti mobilic. Ponte direttod. Riparazione di protesi fratturate

Effetti secondari:
In rari casi i componenti di Super-Bond Brush-dip Starter Kit possono condurre alla
sensibilizzazione. In tali casi Super-Bond Brush-dip Starter Kit non dovrebbe
essere più usato.
2. Componenti
Super-Bond Brush-dip Starter Kit*5 ------------------------ Classe IIa]
Componenti

Quantitativo nel kit

1 Super-Bond Brush-dip Initiator

1 × 0.3 mL

2 Super-Bond Brush-dip Monomer

1 × 3.5 mL

3 Super-Bond Polymer (Brush-dip Clear)1 × 1.5 g

4 Enamel Etchant Gel[Classe I

1 × 1 mL]
Quantitativo nel kit

5 Dispensing Stand

1

6 Dispensing Cups

20

8 Brush Tips (Brush-dip L)

20

7 Brush Handle (Bent)

1
5

9 23G Needle Tip

Documenti allegati: Istruzioni, scheda di istruzioni illustrate, Precauzioni* I componenti del kit possono essere acquistati singolarmente.
5

3. Composizioni
1 Super-Bond Brush-dip Initiator:
tri-n-butilborano parzialmente ossidato, Altro.
2 Super-Bond Brush-dip Monomer:
Monomeri di metacrilato, Altro.
3 Super-Bond Polymer (Brush-dip Clear):
Poli metil metacrilato, Altro.
4 Enamel Etchant Gel:
Acido fosforico, Acqua, Addensante, Pigmento, Altro.
1
2

5. Conservazione

1) Condizioni di conservazione
1 Conservare i componenti di Super-Bond Brush-dip Starter Kit a temperatura
ambiente compresa tra 1 e 30°C e al riparo dalla luce. Elevate temperature, altaumidità e luce solare diretta ne riducono la durata di conservazione.
2 Non utilizzare questo prodotto dopo la data di scadenza. Vedere sull’esternodella confezione la data di scadenza.

6. Smaltimento
1 Rimuovere il tappo della siringa e immergere la siringa vuota di Super-Bond

Brush-dip Initiator in acqua per 24 ore o più. Smaltirlo in conformità con lenormative locali.
2 Smaltire i contenitori o le confezioni vuote (tranne Super-Bond Brush-dip Initiator)
in conformità con le normative locali.

Come in ogni trattamento dentale, la costituzione individuale del paziente e irequisiti unici del caso clinico da trattare vanno presi in considerazione
prima di scegliere i materiali e le condizioni per l'uso.
7. Come si usa
7.1 Preparazione della superficieÈ essenziale che tutte le superfici da incollare con Super-Bond Brush-dip Starter
Kit siano adeguatamente preparate. Il metodo di preparazione può variare aseconda della natura dei materiali.
1) Superficie del dente
1 Pulizia
Pulire tutte le superfici da incollare seguendo la solita procedura. Se necessario,
pulire tutte le superfici utilizzando una coppetta / pennello di lucidatura con
pomice senza olio e senza fluoro. Sciacquare accuratamente e asciugare. Si
raccomanda l'isolamento con una diga di gomma o un rotolo di cotone.
2 Applicazione di Enamel Etchant Gel

AccessoriComponenti

atamente ogni traccia di materiale aderito.
Si raccomanda la pulizia ad ultrasuoni; se tuttavia si preferisce ricorrere all’uso di
una soluzione di pulizia è necessario che sia a ph neutro. Usarla alla concentrazione più appropriata osservandone il manuale di istruzioni. Dopo il lavaggio
rimuovere la soluzione residua risciacquandolo abbondantemente con acquacorrente e asciugarlo immediatamente per prevenire la corrosione.
Dopo la pulizia sterilizzare per 15 minuti a vapore alla temperatura di 121°C. La
sterilizzazione deve essere effettuata seguendo le istruzioni fornite dal produttore
dell’apparecchio sterilizzatore.
Dopo la pulizia e la sterilizzazione accertarsi che non vi siano fessurazioni,
graffiature, scolorimenti o altri danneggiamenti o anomalie.
3 Dopo l’uso pulire lo Dispensing Stand con un comune agente di pulizia e disinfettarli rispettando le norme igieniche prevalenti.

2) Durata del prodotto
La durata del prodotto può essere ridotta a seconda delle circostanze d’utilizzo.
Leggere attentamente prima dell'uso tutte le istruzioni fornite.

Controindicazioni:
Pazienti con storie di gravi reazioni allergiche a questo prodotto o ai monomeri
metacrilici e all’acetone.

[Classe IIa

2 Una volta usato il pennello, lavarne accuratamente il manico e rimuovere immedi-

Forma

Liquido trasparente incolore

Liquido trasparente giallo pallido incolore

34Polvere bianca
Gel rosso

4. Precauzioni

4.1 Avvertenze di sicurezza

1 Super-Bond Brush-dip Starter Kit ha come componenti sostanze come imonomeri metacrilati, che possono causare reazioni allergiche. Non vanno usate
su o da persone sensibili agli acrilati, metacrilati o sostanze simili. L'esposizione
a resine non polimerizzate può causare irritazione, dermatite o reazioni infiammatorie ai tessuti (pelle, mucosa, occhi, ecc.) degli individui sensibili. In tal casointerrompere l'uso, lavare la parte con abbondanti quantità d’acqua e consultare
un medico, se necessario.
2 Evitare che i componenti di Super-Bond Brush-dip Starter Kit entrino in contatto
con tessuti molli, pelle od occhi. La parte contaminata deve essere pulita immediatamente con alcool e poi risciacquata accuratamente con abbondanti quantitàd’acqua corrente. Se i componenti entrano negli occhi, risciacquare immediatamente e accuratamente con abbondanti quantità d’acqua corrente e far visitare ilpaziente da un oftalmologo
3 Evitare l’ingestione accidentale di componenti di Super-Bond Brush-dip Starter
Kit durante l'applicazione e il risciacquo. In caso di ingestione, consultare un
medico se necessario.
4 Maneggiare i componenti di Super-Bond Brush-dip Starter Kit in un’area ben
ventilata (ventilare l’area più volte in un’ora). In caso di inalazione del vapore,
uscire all’aperto.
5 Super-Bond Brush-dip Initiator e Super-Bond Brush-dip Monomer sono altamente
volatili e infiammabili. Non conservarli dove possono essere esposti a fiamme
libere.
4.2 Precauzioni comuni
1 Si raccomanda l’uso di una diga di gomma.
2 I dentisti devono indossare guanti dentali.
3 Non mescolare i componenti con quelli di altri materiali.
4 Utilizzare questo prodotto solo per le applicazioni consigliate in queste istruzioni.

4.3 Precauzioni nella manipolazione
1 Una superficie contaminata da olio, sangue, saliva e biofilm abbassa la forza dellegame. Pulire accuratamente il dente e la protesi. Dopo la pulizia, fare attenzionea evitare la ricontaminazione isolando adeguatamente la superficie. L'uso delladiga di gomma è altamente raccomandato.
2 Fluoruro e olio abbassano la forza del legame. Una pasta dentale contenente
fluoruro o olio non va utilizzata per la pulizia della superficie dei denti.
3 L’eugenolo è un inibitore della polimerizzazione. Pertanto basi e cementi
contenenti eugenolo non vanno utilizzati con i componenti del Super-Bond
Brush-dip Starter Kit.
4 L'ipoclorito di sodio è un inibitore della polimerizzazione. Quando si utilizza
Super-Bond Brush-dip Starter Kit sulla superficie dentinale trattata con ipoclorito
di sodio, Super-Bond C&B Green Activator o Dentin Etchant Gel vanno applicati
dopo l'uso dell’ipoclocrito di sodio. La durata del trattamento con ipoclorito disodio deve essere limitata a meno di 15 secondi. Dettagli su come utilizzare
Super-Bond C&B Green Activator e Dentin Etchant Gel sono disponibili nel
documento allegato.
5 Super-Bond Brush-dip Initiator e Super-Bond Brush-dip Monomer sono altamente
volatili. Chiudere bene subito dopo l’uso.
6 Il liquido attivato (miscela di Super-Bond Brush-dip Initiator e Super-Bond
Brush-dip Monomer) si disattiva gradualmente. Utilizzarlo entro 5 minuti dopo lamiscelazione.
7 Super-Bond Brush-dip Starter Kit può scolorarsi se combinato con un cemento
temporaneo contenente acido tannico.
8 I componenti di Super-Bond Brush-dip Starter Kit sono soltanto monouso. Nonrimettere i componenti erogati nel contenitore originale. Smaltire correttamente ilmateriale residuo. Dopo aver usato Super-Bond Brush-dip Starter Kit con latecnica Brush-dip, smaltire l'eventuale Super-Bond Polymer (Brush-dip Clear) in
eccesso rimasta nel piatto.
9 Le Brush Tips e le Dispensing Cups sono monouso. Non riutilizzarle. Gettarle
dopo ogni utilizzo.
10 Quando una preparazione è profonda, vicina alla polpa, proteggere la polpa con
materiale appropriato.
11 Prima di usare gli accessori controllare che non siano fessurati, graffiati, scoloriti
o in altro modo danneggiati, e in caso di anomalie non utilizzarli.
4.4 Precauzioni con Super-Bond Brush-dip Initiator
Se esposto all'aria o all'acqua, Super-Bond Brush-dip Initiator può perdere attività egenerare calore. Attenersi a quanto segue.
1) Condizioni di conservazione
Evitare elevate temperature, alta umidità e la luce solare diretta. Il cambiamento
ripetuto di temperatura può causare l’aspirazione d’aria nella siringa, riducendocosì la durata di conservazione del Super-Bond Brush-dip Initiator.
*Dopo un tempo prolungato di conservazione, la prima goccia del Super-Bond
Brush-dip Initiator può avere meno attività a causa dell'esposizione all'aria(ossigeno), anche se il resto del materiale nel contenitore rimane inalterato. Lo
scarto di una goccia prima dell'uso garantirà la corretta reazione chimica.
*La siringa è di vetro, quindi deve essere maneggiata con attenzione per evitareurti, caduta e altri danni fisici.
2) Dopo l’uso
Girare la vite della siringa due volte in senso antiorario dopo ogni utilizzo perattenuare la pressione nel cilindro di Super-Bond Brush-dip Initiator. (Lapressione accumulata può causare perdite del Super-Bond Brush-dip Initiator o in
alcuni casi può anche rompere la siringa).
3) Chiusura del tappo
Il tappo si muove solo come aperto e chiuso. Richiudere la siringa subito dopo
ogni utilizzo. L'aria disattiva il Super-Bond Brush-dip Initiator. Non lasciare senza
tappo.

1) Togliere il tappo della siringa (rosso) ruotandolo in senso antiorario.
Conservare il tappo della siringa.
2) Installare la Needle Tip sulla siringa e accertarsi che il collegamento tra ledue sia ben saldo.
3) Applicare il prodotto direttamente sulla superficie del dente.
Nota: Fare riferimento alla Tabella 1 per il tempo di applicazione. Esercitare
lentamente una pressione sullo stantuffo della siringa. Si prega dispingere delicatamente lo stantuffo. Il gel può schizzare intorno
quando viene applicata una pressione forte e veloce.
4) Dopo l’uso eliminare la Needle Tip in modo appropriato e sigillare
immediatamente il contenitore con il tappo della siringa per evitare che ilmateriale si essicchi. Non conservare la siringa con la Needle Tip applicata.
Conservarla esclusivamente con il tappo applicato.
Nota: Quando si applicano i Gel direttamente sulla superficie del dente,
coprire l'intera siringa con una protezione usa e getta (ad esempio una
pellicola di plastica) per evitare la contaminazione da saliva o sangue.
Disinfettare la siringa asciugandola con un tampone di garza assorbente
saturo di disinfettante sia prima della copertura che dopo l'uso.
Table1 Tempo di applicazione
SmaltoEnamel Etchant Gel
30 sec2) Superficie composita in metallo, ceramica, zirconio e resina
1 Pulizia e condizionamento
Pulire tutte le superfici da incollare seguendo la solita procedura. Pretrattare e
condizionare la superficie da incollare seguendo le istruzioni specifiche di ogni
materiale e prodotto protesico. Sgrossare la superficie con fresa in diamante osabbiare se necessario.
Note: L’applicazione di Enamel Etchant Gel contribuisce a rimuovere icontaminanti superficiali e può risultare efficace anche un pulitore ad
ultrasuoni.
2 Applicazione di un primer

Applicare un primer alla protesi secondo le istruzioni fornite dal produttore.
Raccomandiamo l'agente di innesco per protesi dentale prodotto da Sun
Medical Co., Ltd.

7.2 Tecnica Brush-dip
Indicazioni cliniche consigliate:
Fissazione diretta di denti mobili, incollaggio di staffe ortodontiche, incollaggio diponte diretto, riparazione di protesi fratturate ecc.
Nota: Per varie preparazioni di superfici, consultare il punto precedente “7.1Preparazione della superficie”.
1) Erogazione di Super-Bond Polymer (Brush-dip Clear)
Mettere due Dispensing Cups sulla Dispensing Stand. Versare un’adeguata
quantità di Super-Bond Polymer (Brush-dip Clear) in una delle prime coppette.
Appiattire la parte superiore della Super-Bond Polymer (Brush-dip Clear) erogata
nella coppetta toccando delicatamente il lato della Dispensing Stand con le ditao dando dei colpetti sul fondo della Dispensing Stand.
2) Preparazione del liquido attivato
Tenere il flacone di Super-Bond Brush-dip Monomer perpendicolare alla
coppetta e dispensare un numero adeguato di gocce nella seconda coppetta.
Tenere la siringa Super-Bond Brush-dip Initiator perpendicolare alla coppetta,erogare il numero esatto di gocce nel monomero girando la vite. Mescolaredelicatamente con la Brush Tip. Questa miscela è chiamata "Liquido attivato". Il
liquido attivato si disattiva gradualmente, utilizzarlo entro 5 minuti dalla
miscelazione.
[Rapporto di miscelazione del liquido attivat]
Liquido Attivato

Super-Bond Brush-dip Monomer

Super-Bond Brush-dip Initiator

8 gocce2 gocce4 gocce1 goccia3) Applicazione del cemento Super-Bond Brush-dip Starter Kit (resina)
Attaccare la Brush Tip alla Brush Handle. Inumidire la Brush Tip con il liquido
attivato, quindi toccare leggermente la Super-Bond Polymer (Brush-dip Clear)
nella coppetta con la punta del pennello. All’estremità del pennello si formerà una
pallina di Super-Bond Polymer (Brush-dip Clear) che può essere prelevata.
Trasferire e applicare la pallina polimerica sulla superficie da incollare. Non
appena tocca la superficie, il polimero si diffonde per formare uno strato cremoso
ed omogeneo di cemento. Se necessario, ripetere la procedura fino a quando
l'intera superficie non è coperta con l'adeguata quantità di cemento.
Nota: Se si ripete la procedura di prelievo, pulire il pennello con la garza tra le sfereche si formano.
Nota: L’applicazione d’uno strato di liquido attivato sulla superficie prima diposizionare il primo strato aiuta Super-Bond Brush-dip Starter Kit a
distribuirsi senza problemi.
4) Posizionamento e rifinitura
Il tempo di polimerizzazione varia a seconda della temperatura ambiente. Il
tempo di polimerizzazione standard a temperatura orale (37℃) è di 5-6 minuti.
Iniziare la finitura e la lucidatura dopo 6 minuti o più. Accertarsi che si sia
raggiunta una polimerizzazione sufficiente.

ES

4.5 Higiene

1 Brush Tips y Dispensing Cups son de un solo uso. No lo reutilice. Deséchelo

después de cada uso.

INSTRUCCIONESIMPORTANTE :LEA ATENTAMENTE TODAS LAS INSTRUCCIONES ANTES UTILIZAR EL PRODUCTO.
CONSERVE ESTE FOLLETO Y CONSÚLTELO PERIÓDICAMENTE.

Cemento a base de resina adhesiva

Super-Bond Brush-dip Starter Kit
SÓLO PARA USO DEL DENTISTA
1. Descripción del Producto
Super-Bond Brush-dip Starter Kit es un cemento de resina adhesiva deautocurado, que contiene 4-META*1, MMA*2, TBB*3 y PMMA*4 (resina de 4-META /MMA-TBB). Se puede usar para cementar todas las prótesis dentales fijas
fabricadas con una amplia gama de materiales dentales.
*1Anhídrido de trimelitato de 4-metacriloiloxietilo*2Metacrilato de metilo*3Tri-n-butil borano*4Polimetilmetacrilato
Uso previsto:
Super-Bond Brush-dip Starter Kit se adhiere al esmalte dental, la dentina, el metal,la cerámica, el circonio y la resina compuesta con el tratamiento previo adecuado
de la superficie.
Indicaciones:
Super-Bond Brush-dip Starter Kit está indicado para los siguientes:
a. Aplicación de ortodoncia
b. Fijación directa de dientes flojos
c. Puente de unión directa
d. Reparación de prótesis fracturadas

Contraindicaciones:
Pacientes que tienen antecedentes de reacciones alérgicas graves a este productoo monómeros de metacrilato y acetona.

Efectos secundarios:
En raros casos, componentes de Super-Bond Brush-dip Starter Kit pueden causar
sensibilización. En tales casos, ya no debe usarse Super-Bond Brush-dip Starter Kit.
2. Componentes
Super-Bond Brush-dip Starter Kit* ------------------------- Class IIa
5[Class IIa]
Componentes

Cantidad o volumen en el Kit
1 × 0.3 mL

1 Super-Bond Brush-dip Initiator

1 × 3.5 mL

2 Super-Bond Brush-dip Monomer

3 Super-Bond Polymer (Brush-dip Clear)1 × 1.5 g

4 Enamel Etchant Gel[Class I

1 × 1 mL]
Accesorios

Cantidad o volumen en el Kit

5 Dispensing Stand

1

6 Dispensing Cups

20

8 Brush Tips (Brush-dip L)

20

7 Brush Handle (Bent)

1
5

9 23G Needle Tip

Documentos adjuntos: Instrucciones, Tarjetas de Instrucciones Pictóricas, Precauciones*5Componentes del kit se puede comprar de forma individual.
3. Composiciones
1 Super-Bond Brush-dip Initiator:
Tri-n-butil borano parcialmente oxidado y otros.
2 Super-Bond Brush-dip Monomer:
Monómeros de metacrilato y otros.
3 Super-Bond Polymer (Brush-dip Clear):
Polimetilmetacrilato y otros.
4 Enamel Etchant Gel:Ácido fosfórico, agua, espesante, pigmento y otros.
Componentes
1
2

Forma

Líquido transparente incoloro

Líquido transparente amarillo pálido incoloro

3
4Polvo blanco
Gel rojo

4. Precauciones

4.1 Precauciones de seguridad
1 Los componentes de Super-Bond Brush-dip Starter Kit contienen sustancias,
tales como monómeros de metacrilato, que pueden causar reacciones alérgicas.
Por tanto, no son para uso en personas sensibles a los acrilatos, metacrilatos omateriales similares. La exposición a resinas sin curar puede causar irritación,
dermatitis o reacciones inflamatorias a los tejidos (piel, mucosa, ojos, etc.) depersonas sensibles. Si esto ocurre, suspenda el uso, enjuague el área conabundante agua y busque la atención médica, si es necesario.
2 Evite el contacto de los componentes de Super-Bond Brush-dip Starter Kit con
tejidos blandos, piel u ojos. El área contaminada debe limpiarse inmediatamente
con alcohol y luego enjuagarse bien con abundante agua corriente. Si estos
componentes entran en ojo, enjuague inmediata y completamente el ojo conabundante agua corriente y haga que el paciente sea examinado por un
oftalmólogo.
3 Evite ingestión accidental durante la aplicación y el enjuague de los componentes
de Super-Bond Brush-dip Starter Kit. Si se ingiere, busque la atención médica, si
es necesario.
4 Maneje los componentes de Super-Bond Brush-dip Starter Kit en un área bien
ventilada (ventilar el área varias veces por hora). En caso de inhalación del vapor,moverse por aire fresco.
5 Super-Bond Brush-dip Initiator y Super-Bond Brush-dip Monomer son altamente
volátiles e inflamables. No almacene donde puedan estar expuestos a llama
abierta.
4.2 Precauciones comunes

1 Se recomienda el uso de dique de goma dental.
2 Los dentistas deben usar guantes dentales.
3 No mezcle los componentes con otros materiales.
4 Utilice este producto solo para la aplicación recomendada en estas instrucciones.

4.3 Precauciones de manejo

1 La superficie contaminada con aceite, sangre, saliva y biopelículas disminuirá lafuerza de adhesión. Limpie bien el diente y la prótesis. Tras la limpieza, tenga
cuidado de no volver a contaminar mediante el aislamiento adecuado de lasuperficie. Es muy recomendable usar un dique de goma.
2 El flúor y aceite disminuirá la fuerza de adhesión. No debe usarse pasta dental
que contenga el flúor o aceite para limpiar la superficie del diente.
3 El eugenol es un inhibidor de polimerización. Por lo tanto, las bases y cementos
que contienen eugenol no deben usarse con componentes de Super-Bond
Brush-dip Starter Kit.
4 El hipoclorito de sodio es un inhibidor de polimerización. Cuando usa Super-Bond
Brush-dip Starter Kit en la superficie de la dentina tratada con hipoclorito desodio, debe aplicarse Super-Bond C&B Green Activator o Dentin Etchant Gel
después de usar el hipoclorito de sodio. La duración de tratamiento con hipoclorito de sodio debe limitarse a menos de 15 segundos. Detalles sobre cómo usar
Super-Bond C&B Green Activator y Dentin Etchant Gel se puede encontrar en eldocumento adjunto.
5 Super-Bond Brush-dip Initiator y Super-Bond Brush-dip Monomer son altamente
volátiles. Vuelva a tapar inmediatamente después de usar.
6 El Líquido Activado (mezcla de Super-Bond Brush-dip Initiator y Super-Bond
Brush-dip Monomer) se desactiva gradualmente. Úselo en menos de 5 minutos
después de mezclar.
7 Super-Bond Brush-dip Starter Kit puede decolorarse si se combina con un
cemento temporal que contenga ácido tánico.
8 Los componentes de Super-Bond Brush-dip Starter Kit son de un solo uso. No
devuelva los componentes vertidos a su envase original. Deseche el material desobra adecuadamente. Después de usar Super-Bond Brush-dip Starter Kit en latécnica de Brush-dip (cepillo), deseche todo el Super-Bond Polymer (Brush-dip
Clear) sobrante que queda en el recipiente.
9 Brush Tips y Dispensing Cups son de un solo uso. No lo reutilice. Deséchelo
después de cada uso.
10 Cuando una preparación es profunda, cerca de la pulpa, proteja la pulpa dentaria
con el material apropiado.
11 Antes de usar los accesorios, revise daños, grietas, arañazos, descoloramiento,
etc. y deje de usarlo si hay alguna anormalidad.
4.4 Precauciones en Super-Bond Brush-dip Initiator
Cuando se expone al aire o al agua, Super-Bond Brush-dip Initiator puede perderactividad y generar calor. Favor de cumplir con los puntos siguientes.
1) Condiciones de almacenamiento
Evite las altas temperaturas, la alta humedad y la luz solar directa. El cambio
repetido de temperatura puede causar que la jeringa aspire aire, acortando así lavida útil de Super-Bond Brush-dip Initiator.
*Después de un almacenamiento prolongado, la primera gota del Super-Bond
Brush-dip Initiator puede tener una actividad baja debido a la exposición al aire(oxígeno), aunque el resto del material en el envase no se ve afectado. Deseche
una gota antes de usar, así garantizará la reacción química adecuada.
*La jeringa está hecha de vidrio, por lo tanto, debe manipularse con cuidado paraevitar golpes, caída y otros daños físicos.
2) Después de su uso
Gire el tornillo de la jeringa dos vueltas en sentido antihorario después de cadauso para aliviar la presión del barril de Super-Bond Brush-dip Initiator. (La presión
acumulada puede causar fuga del Super-Bond Brush-dip Initiator o en algunos
casos romper la jeringa.)3) Cierre de tapa
La tapa simplemente se desliza hacia dentro y fuera. Vuelva a tapar la jeringa
inmediatamente después de cada uso. El aire desactiva el Super-Bond Brush-dip
Initiator. No deje la jeringa sin tapar.

4) Pulizia della siringa
Pulire la punta della siringa con una garza asciutta dopo ogni utilizzo per evitarel'accumulo di residui. La garza utilizzata deve essere impregnata d'acqua per
disattivare completamente Super-Bond Brush-dip Initiator prima dello smaltimento.
L'accumulo del residuo può causare la chiusura del tappo in modo non corretto.

4) Limpieza de jeringa
Limpie la punta de la jeringa con una gasa seca después de cada uso para evitar
la acumulación de residuos. La gasa para la limpieza debe estar saturada con
agua para desactivar por completo el Super-Bond Brush-dip Initiator antes dedesecharlo. La acumulación de residuos puede causar que el encaje de la tapasea inadecuado.

5) Pulire immediatamente il Super-Bond Brush-dip Initiator sversato con
salviette bagnate
Se assorbito da un materiale combustibile, la temperatura può aumentare
abbastanza da causare fumo. Il Super-Bond Brush-dip Initiator sversato oinutilizzato deve essere immediatamente eliminato con una salvietta monouso
UMIDA (non asciutta). Saturare poi accuratamente la salvietta in acqua corrente
per disattivare il Super-Bond Brush-dip Initiator.

5) Limpiar inmediatamente Super-Bond Brush-dip Initiator derramado con
toallas mojadas
Si es absorbido por un material combustible, puede elevar la temperatura lo
suficiente como para provocar una combustión lenta. El Super-Bond Brush-dip
Initiator vertido o no usado debe secarse inmediatamente con una toalla desechable MOJADA (no seca). Luego sature la toalla para desactivar el Super-Bond
Brush-dip Initiator completamente en agua corriente.

6) Problema nell’erogazione del Super-Bond Brush-dip Initiator
Se il Super-Bond Brush-dip Initiator non forma una goccia a causa dell’ostruzione
dell’orifizio o la vite non gira, non forzarlo. Il contenuto può schizzare se la siringa
di vetro si rompe. Contattare il distributore.

6) Problema al exprimir Super-Bond Brush-dip Initiator
Si el Super-Bond Brush-dip Initiator no forma gotas debido al bloqueo en elorificio o al no poder girar el tornillo, no lo fuerce. El contenido puede salpicar sise rompe la jeringa de vidrio. Por favor contacte a su distribuidor.

7) Altro
In caso di qualunque inconveniente, contattare il distributore.

7) Otros
Si hay algún inconveniente, póngase en contacto con su distribuidor.

2 Después de usar un mango de cepillo, lávelo apropiadamente y elimine inmediat-amente cualquier material adherido. Recomendamos la limpieza ultrasónica,
pero si usa una solución de limpieza, use un detergente o solución de pH neutro.
Use a una concentración apropiada. Utilícelo de acuerdo con el manual de
instrucciones para la solución de limpieza. Después de lavar, enjuague lasolución con suficiente agua corriente y séquelo inmediatamente para evitar la
corrosión. Después de limpiar, se realiza la esterilización por vapor a 121ºC
durante 15 minutos. La esterilización debe llevarse a cabo de conformidad conlas instrucciones del fabricante para el uso del esterilizador. Asegúrese de que nohaya daños, grietas, arañazos, descoloramiento, u otras anormalidades despuésde la limpieza y esterilización.
3 Limpie los Dispensing Stand utilizados con agentes de uso común y desinféctelosde acuerdo con las normas higiénicas de uso común.
5. Almacenamiento1) Condiciones de almacenamiento
1 Almacene los componentes de Super-Bond Brush-dip Starter Kit en una
temperatura ambiente, de 1 a 30°C, y en un lugar oscuro. Las altas temperaturas, la alta humedad y la luz solar directa pueden acortar su vida útil.
2 No utilice este producto después de la fecha de vencimiento. Vea el paquete
exterior para la fecha de vencimiento.
2) Vida útil del producto
La vida útil del producto se puede acortar dependiendo de las circunstancias deuso. Lea todas las instrucciones cuidadosamente antes de su uso.

6. Eliminación
1 Retire la tapa de la jeringa y remoje la jeringa vacía de Super-Bond Brush-dip

Initiator en agua 24 horas o más. Deséchelo conforme a las regulaciones locales.

2 Deseche los envases vacíos o envoltorios excepto Super-Bond Brush-dip Initiator

conforme a las regulaciones locales.

Como en cualquier tratamiento dental, la constitución individual del paciente
y el requisito único del caso clínico deben ser considerados antes deseleccionar materiales y condiciones para el uso.
7. Cómo usar
7.1 Preparación de la superficie
Es esencial que todas las superficies a unir con Super-Bond Brush-dip Starter Kit
estén preparadas apropiadamente. El método de preparación puede variar según
la naturaleza de los materiales.
1) Superficie del diente
1 Limpieza
Limpie todas las superficies a unir siguiendo el procedimiento habitual. Si es
necesario, limpie todas las superficies usando una copa / cepillo de pulido con
piedra pómez sin aceite ni fluoruro. Enjuague bien y seque. Se recomienda elaislamiento con dique de goma o rollo de algodón.
2 Aplicación de Enamel Etchant Gel

1) Retire la tapa (roja) de la jeringuilla girándola en sentido contrario a las
agujas del reloj. Guarde la tapa de la jeringuilla.
2) Coloque con seguridad Needle Tip en la jeringa y asegúrese que la jeringa
esté bien conectada.
3) Aplique el producto directamente en la superficie del diente.
Nota: Consulte la Tabla 1 para el tiempo de aplicación.
Aplique lentamente la presión sobre el émbolo de la jeringuilla. Favorde empujar el émbolo suavemente. El gel puede salpicar alrededor
cuando se aplica una presión fuerte y rápida.
4) Deseche Needle Tip apropiadamente después de usar y selle el envase
inmediatamente con la tapa de la jeringa para evitar que se seque elmaterial. No almacene la jeringa con Needle Tip puesto en su lugar.
Almacénelo solo con la tapa de la jeringa.
Nota: Al aplicar los Geles directamente a la superficie del diente, cubra toda lajeringuilla con una barrera desechable (p. ej. un film plástico) para
prevenir contaminación por saliva o sangre. Desinfecte la jeringuilla
frotándola con una gasa absorbente saturada con desinfectante antes
de cubrir y después de uso.
Tabla 1 Tiempo de aplicación
EsmalteEnamel Etchant Gel
30 seg.

2) Superficie compuesta de metal, cerámica, circonio y resina
1 Limpieza y acondicionamiento
Limpie todas las superficies a unir siguiendo el procedimiento habitual. Hagael tratamiento provisional y acondicione la superficie a ser unida, siguiendo las
instrucciones específicas de cada material protésico y producto. Raspe lasuperficie con fresa de diamante o chorro dental si es necesario.
Nota: La aplicación de Enamel Etchant Gel ayudará a eliminar contaminantes
superficiales, y el limpiador ultrasónico también es eficaz.
2 Aplicación de imprimación

Prepare la prótesis si así lo indican las instrucciones del fabricante .
Recomendamos el uso de agentes de imprimación para prótesis dental
fabricados por Sun Medical Co., Ltd.

7.2 Técnica de Brush-dip (Cepillo)
Casos clínicos recomendados:
Fijación directa de dientes flojos, soportes de ortodoncia de unión, puente de unión
directa, reparación de prótesis fracturadas, etc.
Nota: Para diversos tipos de preparaciones de superficie, favor de consultar elapartado anterior “7.1 Preparación de la superficie”.
1) Dispensar Super-Bond Polymer (Brush-dip Clear)
Coloque dos tazas Dispensing Cups en el plato Dispensing Stand. Vierta una
cantidad apropiada de Super-Bond Polymer (Brush-dip Clear) en una de lastazas. Aplane la parte superior del Super-Bond Polymer (Brush-dip Clear)vertido en la taza golpeando suavemente el costado del plato Dispensing Standcon los dedos o golpeando la mesa por la parte inferior del Dispensing Stand.
2) Preparación de Líquido Activado
Sostenga la botella de Super-Bond Brush-dip Monomer perpendicular a la taza yvierta el número apropiado de gotas en la taza. Sostenga la jeringa deSuper-Bond Brush-dip Initiator perpendicular a la taza y vierta el número
adecuado de gotas en el monómero, girando el tornillo. Remueva suavemente
con el Brush Tip. Esta mezcla se llama “Líquido Activado”. El Líquido Activado
se desactiva gradualmente, úselo en menos de 5 minutos después de la mezcla.
[Proporción de Mezcla para Líquido Activado]
Líquido Activado

Super-Bond Brush-dip Monomer

Super-Bond Brush-dip Initiator

8 gotas2 gotas4 gotas

1 gota3) Aplicación del cemento Super-Bond Brush-dip Starter Kit (resina)
Coloque Brush Tip en Brush Handle. Moje Brush Tip con el Activated Liquid,luego toque el Super-Bond Polymer (Brush-dip Clear) en la taza ligeramente conla punta del cepillo. Una pequeña bola de Super-Bond Polymer (Brush-dip Clear)se formará a la punta del cepillo y puede recogerse. Transfiera y aplique la bolade polímero sobre la superficie a unir. Tan pronto como toque la superficie, el
polímero se extenderá para formar una capa de cemento cremosa yhomogénea. Si es necesario, repita el procedimiento hasta cubrir la superficie
entera con la cantidad adecuada de cemento.
Nota: Cuando repite el procedimiento de recogida, limpie el cepillo con una gasa
entre las bolas formadas.
Nota: Aplicar una capa del líquido activado a la superficie antes de colocar laprimera capa ayudará a extenderse Super-Bond Brush-dip Starter Kit
suavemente.
4) Montaje y acabado
El tiempo de curado varía con la temperatura ambiente. El tiempo de curado
estándar a la temperatura de la boca (37ºC) es de 5 a 6 minutos. Comience elacabado y pulido después de 6 minutos o más. Confirme si se ha logrado el
curado suficiente.

TRTALİMATLAR

ÖNEMLİ :KULLANMADAN ÖNCE TÜM TALİMATLARI OKUYUN.
KULLANIM KILAVUZUNU SAKLAYIN VE PERİYODİK OLARAK OKUYUN.

Dental adhesiv rezin siman

Super-Bond Brush-dip Starter Kit
SADECE DİŞ HEKİMİ KULLANIMI İÇİNDİR
1. Ürüne Genel Bakış
Super-Bond Brush-dip Starter Kit self cure adeziv rezin simandır.
4-META*1,MMA*2,TBB*3ve PMMA*4 (4-META / MMA-TBB rezin) içermektedir. Çokçeşitli materyallerden üretilen tüm sabit protezlerin simantasyonunda
kullanılmaktadır.
*14- metakriloiloksietil trimellitat anhidrit*2Metil metakrilat*3Tri-n-bütil boran*4Poli methil metakrilat
Kullanım amacı:Super-Bond Brush-dip Starter Kit, uygun yüzey ön işlemiyle ; diş minesine, metal,
seramik, zirkonyum ve rezin kompozite bağlanmayı amaçlar.
Endikasyonları:Super-Bond Brush-dip Starter Kit aşağıdaki vakalarda endikedir.
a. Ortodontik uygulamalarb. Mobil dişlerin direk fiksasyonu
c. Direk köprü simantasyonu
d. Kırık protez tamiri
Kontraendikasyonları:
Metakrilat monomerlerine
kontraendikedir.

veya

asetona

alerjiöyküsü

veya daha uzun bir süre suda bekletin. Yerel düzenlemelere uygun olarak atın.
yönetmeliklere uygun olarak atın.

olan

hastalarda]
Kit içindeki miktarı
1 × 0.3 mL

2 Super-Bond Brush-dip Monomer

1 × 3.5 mL

3 Super-Bond Polymer (Brush-dip Clear)1 × 1.5 g

4 Enamel Etchant Gel[Class I

1 × 1 mL]
Aksesuarlar

Kit içindeki miktarı

5 Dispensing Stand

1

6 Dispensing Cups

20

8 Brush Tips (Brush-dip L)

20

7 Brush Handle (Bent)

1
5

9 23G Needle Tip

Ekli dökümanlar : Talimatlar, resimli kulllanım kartı, Önlemler.
*5Kit içeriği ayrıca satılmaktadır.
3. İçeriği
1 Super-Bond Brush-dip Initiator:
Kısmen oksidize tri-n-bütil boran, Diğerleri.
2 Super-Bond Brush-dip Monomer:
Metakrilat monomerleri, Diğerleri.
3 Super-Bond Polymer (Brush-dip Clear):
Poli metil metakrilat, Diğerleri.
4 Enamel Etchant Gel:
Fosforik asit, su, koyulaştırıcı, pigment, Diğerleri.
Bileşenler
1
2

Form

Renksiz şeffaf likit

Renksizden açık sarıya şeffaf likit

342) Ürün ÖmrüKullanım şartlarına bağlı olarak ürünün raf ömrü kısalabilir. Kullanım öncesinde
tüm talimatları dikkatlice okuyun.

2 Boş ürün kaplarını veya ambalajlarını (Super-Bond Brush-dip Initiator hariç) yerel

Super-Bond Brush-dip Starter Kit*5 ------------------------- Class IIaBileşenler

1) Saklama Koşulları
1 Super-Bond Brush-dip Starter Kit bileşenlerini 1-30°C arasında oda sıcaklığında
ve karanlık bir yerde saklayın. Yüksek sıcaklık, yüksek nem ve doğrudan güneş
ışığına maruz kalması ürünün raf ömrünü kısaltacaktır.
2 Son kullanım tarihi geçen ürünleri kullanmayın.Son kullanma tarihi ürün ambalajının dış kısmında yazılıdır.

1 Şırınga kapağını çıkartın. Boş Super-Bond Brush-dip Initiator şırıngasını 24 saat

2. Bileşenleri

1 Super-Bond Brush-dip Initiator

5. Saklama

6. İmha edilmesi

Yan etkiler:Çok nadir vakalarda Super-Bond Brush-dip Starter Kit bileşenleri hassasiyete neden
olabilir. Bu durumlarda Super-Bond Brush-dip Starter Kit kullanımı bırakılmalıdır.

[Class IIa

temizleme solüsyonu için hazırlanmış kullanım talimatlarına uygun olarak
kullanın. Yıkadıktan sonra, musluğun altında yeterince suyla solüsyonu durulayınve paslanmasını önlemek için derhal kurulayın.
Temizledikten sonra, 15 dakika boyunca 121°C’de buhar sterilizasyonunu
gerçekleştirin.
Sterilizasyon, üreticinin sterilizatör kullanımı için hazırladığı kullanım talimatlarınauygun olarak gerçekleştirilmelidir.
Temizleme ve sterilizasyon sonrasında bir hasar, çatlak, çizik, renginde değişiklik
veya başka anormallikler olmadığından emin olun.
3 Kullanılmış Dispensing Stand’yı yaygın olarak kullanılan ajanlarla temizleyin ve
onları yaygın bir şekilde kullanılan hijyen kurallarına uygun olarak dezenfekteedin.

Beyaz toz

Kırmızı jel

4. Önlemler

4.1 Güvenlik Önlemleri

1 Metakrilat monomerleri gibi Super-Bond Brush-dip Starter Kit bileşenleri alerjik

reaksiyonlara neden olabilir. Akrilat, metakrilat veya benzer materyallere hassasiyeti olan kişilerde kullanılmamalıdır. Polimerize olmayan rezinlere temas hassaskişilerde irritasyon, dermatit veya dokularda (deri, mukoza, göz vb.) inflamatuar
reaksiyonlara neden olabilmektedir. Böyle bir durumda, ürünü kullanmayı bırakın,
alanı bol miktarda suyla yıkayın ve gerekirse tıbbi yardım alın.
2 Super-Bond Brush-dip Starter Kit bileşenlerinin yumuşak doku, deri veya göz ile
temasından kaçının. Temas halinde derhal alkol ile silinmeli ve ardından bol
miktarda akan su ile iyice yıkanmalıdır. Bileşenlerin göze temas etmesi durumunda, derhal ve bol miktarda suyla yıkayın ve hastanın bir göz doktoru tarafından
muayene edilmesini sağlayın.
3 Super-Bond Brush-dip Starter Kit bileşenlerini uygularken veya yıkarken yanlışlıkla yutmaktan kaçının. Yutulması durumunda gerekirse tıbbi yardım alın.
4 Super-Bond Brush-dip Starter Kit bileşenlerini iyi havalandırılan bir ortamda
muhafaza edin. (Saatte birkaç kez havalandırın) Buharın solunması durumunda
temiz hava alın.
5 Super-Bond Brush-dip Initiator ve Super-Bond Brush-dip Monomer oldukça
uçucu ve yanıcı materyallerdir. Açık aleve maruz kalabilecekleri bir yerde saklamayın.
4.2 Ortak Önlemler
1 Rubber dam kullanımı tavsiye edilmektedir.
2 Diş hekimlerinin dental eldiven kullanması tavsiye edilmektedir.
3 Bileşenleri diğer malzemeler ile karıştırmayın.
4 Bu ürünü sadece bu talimatlarda önerilen endikasyonlar için kullanın.
4.3 Kullanım için Önlemler
1 Yağ, kan, tükürük ve biyofilm ile kontamine yüzeyler bağlantı kuvvetini düşürmektedir. Dişi ve protezi iyice yıkayın. Yıkama sonrasında yüzeyi yeterince izole
ederek tekrar kontamine olmasını engelleyin.Rubber dam kullanılması önemle
tavsiye edilir.
2 Flor ve yağ ,bağlantı kuvvetini düşüreceği için flor veya yağ içeren diş macunlarının diş yüzeyi temizlenmesinde kullanılması tavsiye edilmez.
3 Öjenol bir polimerizasyon başlatıcısı olduğu için, öjenol içeren baz ve simanların
Super-Bond Brush-dip Starter Kit bileşenleri ile kullanımı tavsiye edilmez.
4 Sodyum hipoklorit bir polimerizasyon başlatıcısıdır. Sodyum hipoklorit uygulanmış
dentin yüzeylerinde Super-Bond Brush-dip Starter Kit kullanılırken, sodium
hipoklorit kullanımı akabinde Super-Bond Green Activator veya Dentin EtchantGel uygulanmalıdır. Sodyum hipoklorit uygulama süresi 15 saniyeden az
olmalıdır. Super-Bond C&B Green Activator ve Dentin Etchant Gel sodyum
hipoklorit kullanım detayları ekli dökümanda mevcuttur.
5 Super-Bond Brush-dip Initiator ve Super-Bond Brush-dip Monomer oldukça
uçucu ürünler olduğu için kullanım akabinde hemen kapağının kapatılması
gerekir.
6 Aktif likit (Super-Bond Brush-dip Initiator ve Super-Bond Brush-dip Monomer)yavaş yavaş deaktive olur. Karıştırma akabinde 5 dakika içerisinde kullanılmalıdır.
7 Super-Bond Brush-dip Starter Kit tanik asit içeren geçici bir siman ile birliştiğinde
rengi daha açık bir renk olabilir.
8 Alınan materyalleri tekrar kabına koymayın. Artan malzemeyi uygun şekilde atın.
Super-Bond Brush-dip Tekniği ile kullanıldı ise uygulama kabında kalan fazla
Super-Bond Polymer (Brush-dip Clear) atın.
9 Brush Tips ve Dispensing Cups tek kullanımlıktır. Tekrar kullanmayın.Kullanım
sonrasında atın.
10 Derin, pulpaya yakın preperasyonlarda, pulpa uygun materyal ile korunmalıdır.
11 Aksesuarları kullanmadan önce hasar, çatlaklar, çizikler, renk değişikliği vb. olup
olmadığını kontrol edin ve herhangi bir anormallik varsa bunları kullanmayı
bırakın.

Tüm diş tedavilerinde olduğu gibi, hastanın kişisel yapısı ve klinik vakanın
kendine has gereksinimleri malzeme ve kullanım koşullarının seçiminde gözönünde bulundurulmalıdır.
7. Kullanımı
7.1 Yüzey Hazırlığı
Super-Bond Brush-dip Starter Kit ile simante edilecek yüzeylerin uygun şekilde
hazırlanması gerekmektedir. Yüzey hazırlığı materyalin doğasına uygun olarakdeğişiklik göstermektedir.
1) Diş Yüzeyi
1 Temizleme
Normal prosedürüne uygun olarak tüm simante edilecek yüzeyleri temizleyin.
Gerekli olması durumunda yağ içermeyen parlatma lastiği / fırçası, florüriçermeyen pomza taşı ile yüzeyleri temizleyin. İyice yıkayıp kurutun. Rubber
dam veya koton pellet ile izolasyon tavsiye edilmektedir.
2 Enamel Etchant Gel Kullanımı

1) Şırınga kapağını (kırmızı) saatin tersi yönünde çevirerek çıkarın. Şırıngakapağını saklayın.
2) Needle Tip’i şırıngaya düzgün bir şekilde takın ve Needle Tip’in şırıngayasıkı bir şekilde oturduğundan emin olun.
3) Ürünü doğrudan diş yüzeyine uygulayın.
Not: Uygulama süresi için Tablo 1'e bakınız.
Şırınga pistonuna yavaşça basınç uygulayın. Lütfen pistonu yavaşça
itin. Jel, güçlü ve hızlı basınç uygulandığında etrafa sıçrayabilir.
4) Needle Tip’i kullandıktan sonra uygun bir şekilde imha edin ve materyalin
tükenmesini önlemek için derhal şırıngayı şırınga kapağı ile kapatın.
Şırıngayı Needle Tip takılı şekilde bırakmayın, sadece şırınga kapağı kapalı
olacak şekilde saklayın.
Not: Jelleri doğrudan diş yüzeyine uygularken, tükürük veya kandan kirlenmeyi
önlemek için tüm şırıngayı tek kullanımlık bir bariyerle (örneğin plastik birfilm) örtün. Kullanmadan önce ve kullandıktan sonra dezenfektanlı bir
gazlı bez ile şırıngayı silerek dezenfekte edin.
Tablo 1 Uygulama Süresi
Mine

Enamel Etchant Gel
30 sn2) Metal, seramik, zirkonyum ve rezin kompozit yüzeylerde kullanımı
1 Temizleme ve Yüzey hazırlama
Normal prosedürüne uygun olarak tüm simante edilecek yüzeyleri temizleyin.
Her bir protez materyali veya ürününün özel kullanım talimatlarına uygun
olarak simante edilecek yüzeyi ön işlemden geçirin ve yüzeyi hazırlayın.
Gerekli olması durumunda yüzeyi elmas frez veya kumlama yöntemi ile
pürüzlendirin.
Not: Enamel Etchant Gel yüzeydeki kontaminatların temizlenmesine yardımcı
olacaktır. Ultrasonik temizleme cihazları da ayrıca etkili olacaktır.
2 Primer Uygulanması

Primer, kullanım talimatında belirtilmiş ise kullanılır.Sun Medical Co., Ltd.
tarafından üretilen diş protez astarlama ajanı öneririz.

7.2 Brush-dip tekniği
Tavsiye edilen klinik vakalar:
Mobil dişlerin direk fiksasyonu, ortodontik braket yapıştırılması, direk köprü
simantasyonu, kırık protez tamirleri, vb.
Not: Farklı yüzey hazırlıkları için, 7.1 Yüzey hazırlama kısmını okuyun.
1) Super-Bond Polymer (Brush-dip Clear)’ın Kaba Alınması
2 adet Dispensing Cups’ı Dispensing Stand’a alın. Uygun miktarda Super-Bond
Polymer (Brush-dip Clear)’ı ilk Dispensing Cups’a alın. Super-Bond Polymer(Brush-dip Clear) kabının kenarına parmaklarınızla hafifçe vurarak veya
Dispensing Cups’ın altını masaya vurarak alınan Super-Bond Polymer(Brush-dip Clear)’ı düzleştirin.
2) Aktif Likidin Hazırlanması
Super-Bond Brush-dip Monomer şişesini dik olarak tutun ve gerekli miktarda
damlayı ikinci kaba alın. Super-Bond Brush-dip Initiator şırıngasını da dik olarak
tutun ve gerekli miktar kadar Super-Bond Brush-dip Initiator’yi Super-Bond
Brush-dip Monomer’in üzerine alın. Brush Tips fırçalar ile karıştırın. Karışım
“Aktif Likit” olarak adlandırılır. Aktif likit zamanla deaktive olacağı için karıştırma
akabinde 5 dakika içerisinde kullanın.
[Aktif Likit karışım oranı]
Aktif Likit

Super-Bond Brush-dip Monomer

Super-Bond Brush-dip Initiator

8 damla2 damla4 damla1 damla3) Super-Bond Brush-dip Starter Kit (rezin siman) uygulanması
Brush Tips fırça ucunu Brush Handle’a (el aletine) takın. Brush Tips fırça ucunu
aktif likit ile ıslatın ve Super-Bond Polymer (Brush-dip Clear) ‘a hafifçe
dokundurun. Fırçanın ucunda küçük bir Super-Bond Polymer (Brush-dip Clear)
topu oluşacaktır ve bu toz topu fırça ile alınabilecektir.Yüzeye temas ettiği an
itibari ile polimer kremsi, homojen bir siman katmanı oluşturmak sureti iledağılacaktır. Gerekli olması durumunda prosedür tüm yüzey kaplanıncaya kadar
tekrar edebilir.
Not: Prosedürü tekrar ederken her toz topu oluşturulması öncesinde fırça ucunu
gazlı bez ile temizleyin.
Not: Toz topunu uygulamadan önce aktif likidin yüzeye ilk katman olarak
uygulanması Super-Bond Brush-dip Starter Kit’un düzgünce dağılmasınayardımcı olacaktır.
4) Sertleşme ve Bitim
Sertleşme süresi ortamın sıcaklığına göre değişiklik göstermektedir. Ağızdaki(37°C) standart sertleşme süresi 5-6 dakikadır. Bitim ve cilalamaya 6 dakika veya
biraz daha sonra başlayabilirsiniz. Yeterli sertleşme sağlandı mı diye kontrol edin.

4.4 Super-Bond Brush-dip Initiator için Önlemler
Hava veya su ile temasında, Super-Bond Brush-dip Initiator aktivitesini kaybedebilirve ısı oluşturabilir. Aşağıda verilen kurallara uyunuz.
1) Saklama koşulları
Yüksek sıcaklıklara, yüksek neme ve direk gün ışığına maruziyetten kaçının.
Sıcaklıktaki tekrarlayan değişiklikler, şırınganın havayı emmesine neden olarak
katalizörün raf ömrünü kısaltabilir.
*Uzun süreli depolamadan sonra, Super-Bond Brush-dip Initiator ürününün ilk
damlası hava (oksijen) ile temas ettiği için aktivitesi daha düşük olabilir. Bununla
beraber kalan materyal etkilenmeyecektir. Kullanmadan önce ilk damlanın
kullanılmadan atılması uygun kimyasal reaksiyonu sağlayacaktır.
*Şırınga camdan imal edilmiştir, bu nedenle şok, düşme ve diğer fiziksel hasarlarıönlemek için dikkatlice kullanılmalıdır.
2) Kullanım sonrası
Super-Bond Brush-dip Initiator şırıngasının basıncını azaltmak için her kullanım
sonrasında vidası saat yönünün tersine 2 tur çevirilmelidir. (Yüksek basınçSuper-Bond Brush-dip Initiator’in sızıntı yapmasına veya bazı durumlardaşırınganın kırılmasına neden olabilir.)
3) Kapağının kapatılması
Şırınga kapağı kolaylıkla açılıp kapatılmaktadır. Her kullanım sonrasında şırıngayı
hemen tekrar kapatın. Hava, Super-Bond Brush-dip Initiator’i deaktive edecektir.
Şırınga kapağını açık bırakmayın.
4) Şırınganın temizlenmesi
Şırınga ucunda kalıntıların birikmesini engellemek için her kullanımdan sonraşırınganın ucunu kuru bir gazlı bez ile silin. Kullanılan gazlı bezin Super-Bond
Brush-dip Initiator’in etkisinden arındırılması için atılmadan önce suda bekletilmesi
gerekmektedir. Şırınga ucunda kalıntının birikmesi kapağının yalnış şekilde
yuvasına oturtulmasına neden olacaktır.
5) Dökülen Super-Bond Brush-dip Initiator’in ıslak havlu ile hemen temizlenmesi
Dökülen ürünün yanıcı bir madde tarafından emilmesi durumunda yanmaya
neden olacak sıcaklığa kadar yükselmesine neden olabilir.Dökülen veya kullanılmayan Super-Bond Brush-dip Initiator hemen ıslak (kuru olmayan) kulan-at bir
mendil ile kurutulmalıdır. Akabinde Super-Bond Brush-dip Initiator’in deaktivasyonu
için akan suda iyice yıkanmalıdır.
6) Super-Bond Brush-dip Initiator ile ilgili bir problem oluşması halinde
Super-Bond Brush-dip Initiator şırınga ağzının tıkanması veya vidanın dönmemesi durumunda şırıngayı zorlamayın. Cam şırınga kabı kırılırsa içerik etrafa
sıçrayabilir. Böyle bir durumda distribütörünüz ile irtibata geçin.
7) Diğerleri
Herhangi bir problem ile karşılaşılması durumunda distribütörünüz ile irtibata
geçin.
4.5 Hijyen

1 Brush Tips ve Dispensing Cups sadece tek kullanımlıktır. Tekrar kullanmayın. Her

bir kullanımdan sonra imha edin.

When any serious incident has occurred in relation to the device, please report to the following
manufacturer and the competent authority in your country.
EMERGO EUROPE B.V.

2 Brush Handle’ı (el aletini) kullandıktan sonra güzelce yıkayın ve üzerine yapışmış

herhangi bir materyal bulunuyorsa derhal maddeyi çıkartın.
Ultrasonik temizleme yapılması tavsiye edilmektedir fakat eğer bir temizleme
solüsyonu kullanıyorsanız, nötr pH değerli bir temizleyici ya da solüsyon
kullanmaya dikkat edin. Temizleyiciyi veya solüsyonu uygun bir yoğunlukta ve

Westervoortsedijk 60
6827 AT Arnhem The Netherlands

Date of issue: 2022-12-22

J029/2F