3.2 To get the best results
qAQ Bond Plus Sponge Activators/Brushes
AQ Bond Plus Sponge Activators/Brushes contain a bonding
promoter and are an important component of AQ Bond Plus. Always
apply AQ Bond Plus using AQ Bond Plus Sponge Activators/Brushes and do NOT substitute any other sponges or brushes.
wStorage
Store AQ Bond Plus in a dark location between 4˚C and 15˚C,
preferably in a refrigerator, but do NOT freeze AQ Bond Plus. If
frozen, phase separation will occur. High temperature, direct
sunlight and high humidity should be avoided. AQ Bond Plus
Sponge Activators/Brushes are slightly hydroscopic.
eVolatility
Base is highly volatile. Firmly recap the bottle immediately
after use.
rContamination
To avoid contamination, do not mix the components with those ofother materials, nor mix the bottle caps. Particularly, do not mix
AQ Bond Plus Sponge Activators/Brushes with other sponges orbrushes. After use, dispose of leftover material. Once dispensed,
do not return the materials to their original container.

INSTRUCTIONSIMPORTANT :READ ALL INSTRUCTIONS THOROUGHLY BEFORE USE.
KEEP THIS LEAFLET AND REFER TO IT PERIODICALLY.

Bonding Agent for Resin-based Dental Restorations and Protective Coat for Dentin

ABond Plus

FOR DENTIST USE ONLY

1. What is AQ Bond Plus?
AQ Bond Plus is a single-bottle, self-etching, self-priming, light-curing
dentin bonding system. It does not require acid-etching of dentin or
cut enamel surfaces before use.* It is suitable for use with directly
applied composite resins and can be cured by any light-curing
devices. AQ Bond Plus can also be used as the adhesive agent forindirectly fabricated restorations. It contains 4-META**, a highperformance adhesive monomer that decalcifies tooth substrate andpenetrates through the smear layer to form a hybrid layer, creating
excellent bonds to dentin and enamel. It forms a fortified resin coaton dentin, which blocks mechanical and biological stimuli todentin-pulp-complex.
* See "Note" in section 4.q below** 4-methacryloxyethyl trimellitate anhydride

2. Contents of the kit:
qBase ---------------------------------------------------------- 5.5mL
wAQ Bond Plus Sponge Activators -----1 (approx. 175 pcs)
ePlastic Dispensing Dishes ----------------------------------- 15
Sold separately:• Base ----------------------------------------------------------- 5.5mL• AQ Bond Plus Sponge Activators ------1 (approx. 175pcs)• AQ Bond Plus Brushes ------------------ 1 box (100 brushes)• Plastic Dispensing Dishes ------------------------------------ 303. Precautions: Read all instructions thoroughly before use.
3.1 Safety
qSensitivity
AQ Bond Plus should not be used by clinicians or on patients
who are methacrylate monomer-sensitive. If signs of irritationsuch as redness appear, stop using it immediately and consult aphysician.
wPrecautions
AQ Bond Plus contains acidic methacrylate monomer. Avoid
contact of the AQ Bond Plus components with soft tissue, skin oreyes. Use of rubber dam is recommended. Contaminated skin or
mucosa should be wiped off immediately with alcohol and thenrinsed thoroughly with copious amounts of running water. Insome cases contaminated mucosa or gingiva may whiten, which
will usually disappear in a few days. To minimize contamination
during blowing or coating, place an evacuator near the treated
tooth. If AQ Bond Plus Sponge Activators/Brushes contact wet
skin, the odor may remain. If the AQ Bond Plus components
enter the eye, immediately rinse the eye thoroughly with runningwater. Then the patient should be examined by an ophthalmologist.
Clinicians should use dental protective gloves.
Base contains approx. 40% acetone. Inhalation of a large amountof concentrated vapor may cause headache: therefore the area
should be well ventilated. In case of such severe inhalation, moveto fresh air.
ePulp protection
If the preparation approaches the pulp, apply a protective liner orbase.
rFlammability
Base is flammable. AVOID OPEN FLAME in use and storage.
tApplications
Use AQ Bond Plus only for the applications recommended in theinstructions.

As in any dental treatment, the patient's individual constitution
and the unique requirements of clinical case at hand must be
considered before selecting materials and conditions for use.
4. How to use AQ Bond PlusqSurface Preparation
Prepare tooth using an instrument such as fine-grit point so that thesmear layer becomes minimal. After tooth preparation, wash andblow-dry the surface.
Note: The bond strength to enamel can be further improved by
pretreatment of enamel for 5 – 10 seconds with a phosphoric
etchant such as Super-Bond C&B Red Activator/Enamel
Etchant Gel but do NOT use silica- thickened phosphoric
etchant which will compromise the bond strength.
Fluorinated enamel should be always etched as above.
wApplication of the Basea)Dispensation• Dispense one drop (up to 2 drops for a larger surface) of Basein the well of a Plastic Dispensing Dish and firmly recap thebottle immediately.
• Stir the expressed liquid in the well with a AQ Bond Plus
Sponge Activator/Brush for a few seconds.
Note: The liquid in the mixing well should be used within 3
minutes to minimize evaporation.
b)Application• After stirring with the AQ Bond Plus Sponge Activator/Brush,
apply onto the prepared tooth surfaces and keep it moist for 20seconds.
• Evaporate the solvent using a very gentle air blow for 5 - 10
seconds, then another 5 - 10 seconds to dry the surface using
strong air.

eLight-curing
Irradiate light with a visible light curing unit. With an ordinary
halogen lamp irradiate 5 seconds for direct restorations, and 10
seconds for indirect restorations.
Note: AQ Bond Plus is compatible with any currently available
curing lights. With a fast light, the irradiation time should beshortened according to the instruction of the manufacturer
and reflecting the above shorter irradiation time for direct
restorations.
rRestorationa)Direct restoration with a composite
Place a composite and light-cure according to the manufacturer'sinstructions. For deep cavities use incremental composite
placement and, where applicable, line with a flowable resin.

b)Indirect restoration• Second coating is recommended to avoid the hypersensitivity.
(OR/ Repeating the application of AQ Bond Plus makes thecoat stronger.)• Wipe the surface of the light-cured AQ Bond Plus with an
alcohol cotton pellet to remove the air-inhibited layer. Wherenecessary, re-shape the margin.
• Take an impression using a material of your choice.
• Temporize and fabricate the prosthesis. If a temporary resin isused, apply a separating agent beforehand to avoid inadvertent
bonding to the tooth.
• Remove the temporary, wash and dry the cavity.
• Apply a luting agent of your choice and place the prosthesis.

도포하는 경우 항상 AQ Bond Plus Sponge Activators/Brushes를사용하고 다른 Sponge 나 Brush로 대용해서는 안된다.
w보관A Q B o n d P l u s 를 가 능 한 냉 장 고 나 4～15℃ 의 어 두 운 장 소 에보관한다. AQ Bond Plus를 냉동시켜서는 안된다. 냉동시키면 상분리가 일어난다. 고온, 다습, 직사광선을 피할 것.
AQ Bond Plus Sponge Activators/Brushes는 약간의 흡습성이 있다.
e휘발성Base는 휘발성이 강하다. 사용 후에는 곧바로 병을 단단히 밀폐할 것.
r오염오염을 피하기 위해 구성품을 다른 재료의 구성품과 혼용하거나 병마개를 혼용하지 말 것.
특히 AQ Bond Plus Sponge Activators/Brushes를 다른 어떤Sponge/Brush와도 섞어서는 안 된다. 사용 후 남은 재료는 폐기할것. 일단 채취한 재료를 원래의 용기에 담지 말 것.

4. AQ Bond Plus의 사용법q치면형성취급설명서사용 전에 본 취급설명서를 읽어 주십시오。

치과수복용 본딩재 및 상아질 코팅재ABond Plus치과의사전용

１. AQ Bond Plus 란？
AQ Bond Plus는 1 액상에 Etching과 Priming의 기능을 겸비한 광경화형상아질 Bonding System 으로 상아질 또는 절삭 에나멜면의 산 Etching이 필요치 않다*.
직접법으로 충전하는 Composite Resin 의 접착에 적합하고, 어떤 종류의광중합기로도 사용할 수 있다. 또한 AQ Bond Plus 는 간접법으로보철물을 장착할 때 접착제로써도 사용할 수 있다. 고성능 접착성Monomer인 4-META ** 를 함유하여 치질을 탈회하고 건전상아질내에침투하여 수지함침층을 형성함으 로써 상아질과 에나멜에 대한 뛰어난접착성능을 발휘한다.
이 층은 상아질에 강화수지막을 형성하여 상아질·치수복합체에 대한기계적, 생물학적인 자극을 차단한다.
* 하기４.１항의「주의」를 참조할 것.
** 4-methacryloxyethyl trimellitate anhydride

Smear Layer 형성을 최소한으로 하기 위해 fine-grit point 등의기 구 를 사 용 하 여 치 아 의 전 처 리 를 한 다 . 형 성 후 치 면 을 세 정·건조시킨다.
주의：에나멜에 대한 접착력은 Super-Bond C&B Red Activator/
Enamel Etchant Gel 와 같은 인산Etching재로 5~10초간전처리를 하면 더욱 강력해 진다. 단 실리카 함유 인산 Etching재는 접착력을 저하시키므로 사용하지 말 것.
불산 처리한 에나멜에 대해서는 항상 상기의 전처리를 행할 것.

wBase의 도포a)조제
·Plastic Dispensing Dish의 홈에 Base １방울(피착면이 넓은경우에는 ２방울까지）을 떨어뜨리고 곧바로 병 마개를 단단히닫는다.
·Base에 AQ Bond Plus Sponge Activators/Brushes를 넣고 3~4
회 휘젓는다.
주의：혼합된 액재는 휘발을 최소한으로 하기 위하여 3분 이내에사용할 것.
b)도포
·혼합 후 형성면에 AQ Bond Plus Sponge Activators/Brushes로혼합한 혼합물을 도포한다. 형성면을 20초간 액으로 덮인 상태로유지할 것.
·아주 약한 Air Blow를 5~10초간 불어 액재가 고르게 도포되게한다. 그런 뒤 5~10초간 강하게 Air Blow를 하고 표면을건조시킨다.

e광경화
２. Set 구성품：qBase -------------------------------------------------- 5.5mL
wAQ Bond Plus Sponge Activators ----------- 1 (약175개）
ePlastic Dispensing Dishes ------------------------------- 15[별매품]·Base --------------------------------------------------- 5.5mL·AQ Bond Plus Sponge Activators ----------- 1 (약 175개)·AQ Bond Plus Brushes --------------------- 1 box (100개)·Plastic Dispensing Dishes ------------------------------- 30

３. 사용상의 주의：사용전에 잘 읽을 것.
３.１ 안전q과민증상AQ Bond Plus는 메타크릴산 모노머에 과민증이 있는 임상의가사용하거나 과민증이 있는 환자에게 사용하지 말 것. 만약 발진 등의 과민징후가 나타난 경우 곧바로 사용을 중지하고 의사의 진단을 받을 것.

w사용상의 주의AQ Bond Plus는 산성의 메타크릴산 모노머를 배합하고 있다.
구강점막, 피부 또는 눈에 들어가지 않도록 할 것. 러버댐의 사용을
추천한다. 피부 또는 점막에 묻은 경우에는 곧바로 알코올로 닦아내고흐르는 물로 충분히 세정할 것. AQ Bond Plus가 묻은 점막이나치육이 하얗게 되는 경우가 있으나, 보통 며칠 안에 사라진다. Air
Blow 또는 Coating처리시에 주위에 묻는 것을 최소한으로 하기위하여 처치할 치아 주위에 바큠을 둘 것. 만일 AQ Bond Plus
Sponge Activators/Brushes가 피부와 접촉하면 냄새가 남을 수있다. 만일 눈에 들어가면 곧바로 흐르는 물로 충분히 세정하고안과의의 진단을 받을 것. 임상의는 치과용 글로브를 착용할 것.
Base는 약40%의 아세톤을 함유하고 있다. 고농도의 Base 증기를대량으로 흡입하면 두통을 일으킬 가능성이 있으므로 환기를 충분히시킬 것. 다량으로 흡입한 경우 신선한 공기가 있는 장소로 이동할 것.

가시광선조사기로 조사한다. 일반적인 Halogen으로 직접법수복인경우에는 5초간, 간접법수복인 경우에는 10초간 광조사한다.
광원광강도

LED

600-800mW/cm2 약５초조사시간

Halogen 120mW/cm2 이상 약５초
Plasma2,000W/cm2

약３초

주의：AQ Bond Plus는 모든 광조사기로 경화시킬 수 있다. 조사능력이높은 광조사기를 사용할 경우에는 제조업자의 설명서에 따라조사시간을 단축시키고, 직접법수복에서는 상기의 짧은조사시간을 고려할 것.

r수복a)Composite에 따른 직접법수복Composite Resin을 충전하고 제조업자의 설명서에 따라광경화시킨다. 와동이 깊은 경우에는 Composite Resin을적층충전하고 가능하다면 Flowable Resin으로 라이닝할 것.
b)간접법수복
·지각과민방지를 위해 2번 도포를 추천한다.
（AQ Bond Plus를 반복하여 도포함으로써 코팅막을 강화시킬 수있다）·미중합층을 없애기 위하여 알코올솜으로 광경화한 AQ Bond Plus의표면을 닦아낸다. 필요하면 변연부를 재형성한다.
·선호하는 인상재를 사용하여 인상을 채득한다.
·Temporary 보철물을 제작한다. Resin계의 Temporary를 사용할경우 치질에 Resin이 묻지 않도록 미리 분리재를 이용할 것.
·Temporary를 떼어내고 와동을 세정·건조시킨다.
선호하는 루팅재를 이용하여 보철물을 접착한다.

어떠한 치과진료에서든 재료와 사용조건을 선택하기 전에 환자의개인차와 개개의 증례에 따른 요구를 고려할 것.

e치수보호형성면이 치수에 가까운 경우 보호용 라이너 또는 Base를 이용할 것.

r인화성Base는 가연성이므로 사용 중 또는 보관시에 화기에 주의할 것.

t사용범위설명서에 기재되어 있는 사용법대로만 AQ Bond Plus를 사용할 것.
3.2 최고의 효과를 얻기 위하여qAQ Bond Plus Sponge Activators/Brushes
AQ Bond Plus Sponge Activators/Brushes는 접착촉진재를포함하는 AQ Bond Plus의 중요한 구성품이다. AQ Bond Plus를

Prepared/Revised on April 2, 2012

J421E/3D