ULTRAThe homogeneity of a superconducting magnet on your bench
Ultra-high field homogeneity combined with solvent suppression allows you to resolve compounds
dissolved at sub-millimolar concentrations in protonated solvents, such as water.

Resolving metabolites in urine at milli-molar concentrations
State of the art
IN JUST 8MINUTES

Spinsolve ULTRA

ZOOMx 2000

freq. [ppm]Quantifying sugar content in water
Sugars are typically dissolved in water at concentrations that can be very low and most lines overlap
with the large water peak. The Spinsolve ULTRA can be used to resolve even the α and β protons toquantify the amount of glucose in water based products.
By eliminating overlap with the large solvent signals theSpinsolve ULTRA offers high linearity even for integration
regions very close to the solvent peaks

Measuring sugar content in soft drinks
Spinsolve ULTRA measures not only the sugar content but canalso identify the type of sugars present in the sample. The highsensitivity of the system allows you to quantify with high accuracy
the ethanol content typically present at very low concentrations innatural fruit juices. The samples here are all neat. All
measurements took 8 minutes, except Coke Zero was 1 hour.
Sucrose Glucose Fructose Total
Total(NMR) (Label)

Ethanol

Grape

0.0

10.0

6.2

16.2

16.0

0.16

Apple

1.1

2.4

6.4

9.8

9.8

0.03

Orange

3.7

2.0

3.8

9.5

9.0

0.02

Coke

6.0

2.8

1.9

10.7

10.70

Pepsi

1.1

5.6

4.4

11.2

10.7

0

Zero000000*All units g/100ml. Ethanol is % vol.

Detection of metabolites in urine
The spectra of different urine samples show large variations in the concentrations of the various observedmetabolites. The Spinsolve ULTRA can detect sub-millimolar concentrations in a measurement time of just
8 minutes.

Applications
The high homogeneity of the Spinsolve ULTRA is particularly useful for samples where the compounds
to be identified and quantified are dissolved in protonated solvents such as water. Applications include:� Quantification of sugars and alcohol in beverages

� Identification and quantification of metabolites in urine� Analysis of contaminants in waste water� Reaction monitoring in the presence of protonated solvents� qNMR for samples where important signals overlap
� Monitoring fermentation processes in bioreactors

Contact us to see if your application is feasible on a Spinsolve ULTRA.
The high homogeneity of the Spinsolve ULTRA is possible due to advances in the patented shimming
technology used in the Magritek High Homogeneity Halbach Magnets* inside every Spinsolve ULTRAsystem.
*Patent US 8,148,988 and EP 2,144,076

Linewidth measured in a chloroform spectrum
� No sample spinning, no spinning side bands, no requirement for compressed air� No reference deconvolution, only standard NMR data processing is used

freq. [Hz]Spinsolve ULTRA� Available with either 43 MHz or 60 MHz operating frequency (1H)
� Nuclei: All models measure 1H and 19F
Optional X-nuclei: 7Li, 11B, 13C, 15N, 29Si, 31P (ask for other nuclei available)

� Linewidth specifications for all models measured on chloroform
� 50% Linewidth: < 0.2 Hz� 0.55% Linewidth: < 6 Hz

� 0.11% Linewidth: < 12 Hz

� Includes Solvent Suppression Pulse Sequence
Please contact us either by email (sales@magritek.com) or using the phone numbers below to find out
more about Spinsolve ULTRA. We can run feasibility measurements to see if Spinsolve ULTRA is suitablefor your application.

CONTACT INFORMATION

For further information, please contact: sales@magritek.com

GERMANY				NEW ZEALAND												Philipsstraße 8				
6 Hurring Place, Unit 3			52068 Aachen, Germany		
Newlands, Wellington 6037, NZ		Tel: +49 (241) 70525-6000		
Tel: +64 4 477 7096			
Fax: +49 (0)241 539-4375		
Fax: +64 4 471 4665			
Or visit our website www.magritek.com

UNITED STATES		
6440 Lusk Blvd (D108)
San Diego, CA 92121, USA
Tel: +1 (855) 667-6835+1 (866) NMR-MTEK