English

INSTRUCTION MANUALFORDIGITAL BLOOD PRESSURE MONITOR

हिन्दी

Pyсский

Français

Español

CHU304
CHU305EnglishEspañol
FrançaisPyсский
हिन्दी

*P
 lease read all of the information in the instruction
manual before operating the monitor.

00_CHU304_305_2401.indb 1

Rev.2401

8/3/2024 9:21:18 AM

EnglishContents

EnglishEspañol
Français

1 - GENERAL REMARKS
3 - PRECAUTIONS FOR USE AND MAINTENANCE
6 - IDENTIFICATION OF PARTS
7 - LOADING THE BATTERIES
7 - REMOVING THE BATTERIES
8 - TIME ADJUSTMENT
9 - MEASURING YOUR BLOOD PRESSURE
12 - TIPS FOR OBTAINING ACCURATE MEASUREMENTS
12 - CALLING UP THE DATA STORED IN MEMORY
14 - ABOUT “BLOOD PRESSURE”
14 - BLOOD PRESSURE Q&A
15 - BEFORE REQUESTING REPAIRS OR TESTING
16 - SPECIFICATIONSPyсский

■ Ensure that you have all the following components.
• Blood pressure monitor unit • Instruction Manual• 4 AA-size batteries (for monitor)• Cuff (model: SCN-008) for CHU304• Wide Range Cuff (model: SCW-009) for CHU305

GENERAL REMARKS
General remarks on blood pressure and blood pressure measurementहिन्दी

1.
2.

3.
4.
5.
6.

To deflate the cuff quickly, press the “START/STOP” switch.
Before applying your blood pressure cuff, be sure you have selected theappropriate size cuff:
Cuff (model: SCN-008) fits arm circumference: 22 cm – 30 cm.
Wide Range Cuff (model: SCW-009) fits arm circumference: 22 cm – 42 cm.
When attaching the cuff, adjust the position of the cuff so that the air hose is located in
the center of your arm and the hem of the cuff is 1 cm – 2 cm above your elbow.
Self measurement is not therapy! In no event should you change the dosage ofyour medication prescribed by your physician.
As preparation for measuring your blood pressure, you should urinate and thenremain relatively still for 10 to 15 minutes prior to measurement.
DO NOT place too much emphasis on the results of one measurement. Keep acontinuing record of your blood pressure variations. A complete picture can only
be obtained from a large number of readings.

To obtain correct blood pressure measurements
1.
2.

Take five or six deep breaths and then relax before measuring your blood pressure.
If you are tense when taking the measurement, you will not get a valid reading.
Your blood pressure will be elevated if you are anxious or irritated,suffering from lack of sleep or constipation, or have just taken some
exercise or eaten a meal.

00_CHU304_305_2401.indb 1- Eng 1 -8/3/2024 9:21:18 AM

Remarks:
The graph is not exact, but
may be used as a guide inunderstanding non-invasive
blood pressure measurements.
The device is only intended foruse by adults.

Definitions and Classifications of Blood Pressure Levels* This unit is equipped an indicator which visually indicates the blood pressureclassification (Optimal/Normal/High-Normal/Grade 1 Hypertension/Grade 2
Hypertension/Grade 3 Hypertension) of the result after each measurement.
11. Intended Use• This device is noninvasive blood pressure monitor by oscillometric method.
This device can measure the systolic blood pressure (SYS), the diastolic bloodpressure (DIA) and the pulse rate automatically.
•T his device is intended to be used for checking personal health condition at home
under the direction of a physician and is not intended to be a diagnostic device.
• This device is not intended for use with neonates or infant, and not intended
for automatic cycling measurement.

EnglishEspañol

DO NOT try to measure your blood pressure immediately after drinking coffee or
tea or after smoking.
7. Measure your blood pressure when you are relaxed and still. Keep the center ofthe cuff at the level of your heart and DO NOT move your arm or talk.
8. DO NOT measure blood pressure when the cuff has been on your arm for a
few minutes or more. During this time, your arm will have built up a higher bloodpressure and you will not get valid reading.
9. If the Irregular Heartbeat (IHB) Indicator is displayed, relax and measure
your blood pressure again. If the Irregular Heartbeat (IHB) Indicator appears
frequently, you are advised to consult a doctor about your health condition.
See page 11)(
10. The figure below shows the blood pressure classifications under the WHO standards.

Français6.

Pyсский

4.
5.

Wrap the cuff around your arm and tighten it so that one finger can be
See page 9, 10)
inserted between the cuff and arm. (DO NOT measure your blood pressure after bathing or drinking.
Measure your blood pressure where the room temperature is around
20°C. DO NOT measure your blood pressure when it is too cold (below
10°C) or too hot (above 40°C) in the room.

हिन्दी

3.

Measure your blood pressure at the same time each day.

* Your blood pressure changes all the time. This means that data gathered overa long period has far more significance than data from just one measurement.
For this reason, you must measure your blood pressure on a daily basis. Ideally,
you should measure your blood pressure at the same time each day, wherever
possible.

00_CHU304_305_2401.indb 2- Eng 2 -8/3/2024 9:21:18 AM

PRECAUTIONS FOR USE AND MAINTENANCE
Precautions for use
1.

EnglishEspañol
FrançaisPyсский
हिन्दी

If you suffer from heart disease, high blood pressure or other circulatory
disease, consult your physician before using the monitor.
2. Inflating to a higher pressure may result in bruising where the cuff is applied.
If the cuff pressure feels abnormal or you experience any other irregularity while
using the cuff, reduce the pressure immediately by pressing the “START/STOP”
switch, detach the cuff or unplug the air hose connector from the monitor andthen consult the sales outlet where you purchased the monitor.
3. If you think the measurement is abnormal or if measurement makes you feelunwell, discontinue use and consult your physician.
4. Blood pressure measurement may not be possible for anyone with a weakpulse or arrhythmia.
5. Repeated blood pressure measurement and prolonged overinflation may cause
problems such as congestion or swelling in some people.
6. Frequently repeated blood pressure measurements will not give accurate
results. Allow an interval of about 1 minute between measurements.
7. If you suffer from a severe problem with blood circulation in your arms, consult
your physician before using the monitor. Failure to do so could be hazardous toyour health.
8. Measurement may not be possible for anyone with insufficient blood flow to thearea where measurements will be taken or who suffers from a frequent irregularheartbeat. Consult your physician for advice on whether to use the monitor.
9. DO NOT wrap the cuff around an injured arm.
10. DO NOT wrap the cuff around an arm in which a drip (intravenous infusion)is inserted or which is being used for blood transfusion as part of medical
treatment. Doing so could result in an injury or a serious accident.
11. Do not share the cuff with other infective person to avoid cross-infection.
12. If you have any doubt about the application of the cuff on the arm on the side ofa mastectomy, consult your physician.
13. DO NOT use the monitor in the vicinity of flammable gases such as those used
for anaesthesia. Doing so could ignite the gases and cause an explosion.
14. DO NOT use the monitor in enriched oxygen environments such as a hospital’s
hyperbaric chamber or oxygen tent. Doing so could ignite the oxygen andcause a fire.
15. DO NOT use mobile phones near the monitor as this could result in a malfunction.
16. If you use a cardiac pacemaker, consult your physician before using themonitor.
17. Be sure to use this unit only for measuring blood pressure. DO NOT use it for
any other purpose.
18. Be sure to use only our authorized parts and accessories. Parts andaccessories not approved for use with the device may damage the unit.
19. DO NOT use this unit on infants.
20. Blood pressure measurement may not be possible for anyone with common
arrhythmias such as atrial or ventricular premature beats or atrial fibrillation.

00_CHU304_305_2401.indb 3- Eng 3 -8/3/2024 9:21:18 AM

7.
8.
9.

Warning

No modification of this device is allowed.

Symbols Explanation: The CE marking is meant to provide information to market inspectors
in the EU member countries.
: Type BF applied part (The cuff is type BF applied part.): Keep dry: Warning: Refer to instruction manual before use (The sign background color:
blue. The sign graphical symbol: white.): Appliance compliance WEEE directive: Manufacturer: Symbol for “EUROPEAN REPRESENTATION”

Warning

Español5.
6.

Français

3.
4.

Pyсский

2.

DO NOT store the blood pressure monitor in locations exposed to direct
sunlight, high temperatures (over 55°C), low temperatures (below -20°C),
high relative humidity (over 90%) or excessive amounts of dust.
Make sure to store the blood pressure monitor, where children, pets and or
pests are not there.
DO NOT drop the blood pressure monitor or the cuff and subject it to othershocks or vibration.
Remove the batteries if the monitor will be left unused for a long period.
DO NOT attempt to disassemble the monitor or the cuff. Ask the service center
which is designated by manufacturer.
DO NOT bend the cuff or the air hose excessively.
If the monitor and the cuff are very dirty, wipe them clean with a cloth moistenedwith sterilizing alcohol or a neutral detergent. Then wipe them with a dry cloth. If
needed in hospital, the cuff should be disinfected 2 times every week.
NEVER clean the blood pressure monitor with alcohol, thinners or benzene, as
this could damage the monitor.
To clean the cuff, wipe it with a moist cloth. Avoid hard rubbing, as this willcause air leakages. Take care also not to get water into the air hose.
Clean the cuff after the usage of every 200 times is recommended.

हिन्दी

1.

EnglishMaintenance Precautions

The device has been tested and homologated inaccordance with EN60601-1-2 for EMD. This does notguarantee in any way that the device will not be affectedby electromagnetic interference. Avoid using the device inhigh electromagnetic environment.

Nature and frequency of maintenance:

This product is designed for use over an extended period of time;
however, it is generally recommended that it be inspected every two
years to ensure proper function and performance.

00_CHU304_305_2401.indb 4- Eng 4 -8/3/2024 9:21:18 AM

Protect the nature environment:

Please help to protect natural environment by respecting national and/or local recycling
regulations when disposing of the battery and the product at the end of their useful live.

WEEE MARKEnglishIf you want to dispose this product, do not mix with general household
waste. There is a separate collection systems for used electronics products inaccordance with legislation under the WEEE Directive (Directive 2012/19/EU)
and is effective only within European Union.

Information on Disposal in other Countries outside theEuropean Union.

Español

If you wish to discard used batteries, please contact your local authorities or dealer
and ask for the correct method of disposal.

Note for the battery symbol.

The symbol might be used in combination with a chemical symbol. In this case it
complies with the requirement set by the Directive for the chemical involved.

Reference European standard:
Français

The blood pressure measuring device corresponds to regulation EN60601-1, EN10601, EN1060-3, EN1060-4.

Calibration:

Pyсский

The blood pressure measuring device is generally recommended to have themonitor inspected every two years to ensure correct functioning and accuracy.
Please contact a distributor.

हिन्दीName :
Address :Factory
Name :
Address :CITIZEN SYSTEMS JAPAN CO., LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan
ANDON HEALTH CO., LTD.
No.3 Jinping Street, YaAn Road, Nankai District, Tianjin 300190, China

European Representative:
Name :
Address :EMERGO EUROPE
Westervoortsedijk 60, 6827 AT Arnhem, The Netherlands• For technical data, please contact our European
Representative

Made in China

00_CHU304_305_2401.indb 5- Eng 5 -8/3/2024 9:21:18 AM

IDENTIFICATION OF PARTS
Component names
Air hose

Display

MEMORY
SwitchMetal hoop

Air hoseconnector

Français

START/STOP
Switch Stand-by
Air hose socket

Pyсский

* See page 7 for the Battery compartment<Display>
Battery icon

Ready to inflate
indicator

Systolic bloodpressure display
Hypertension
Classification
Indicator

Español

English<Cuff>

हिन्दी

<Main Unit>

Memory icon

Diastolic bloodpressure/Pulse
display

Irregular Heartbeat(IHB) Indicator*

Date/Time display* See page 11 for the Irregular Heartbeat (IHB) Indicator.

00_CHU304_305_2401.indb 6- Eng 6 -8/3/2024 9:21:19 AM

LOADING THE BATTERIES(Batteries supplied with the unit should be loaded in compartment before you use
the blood pressure monitor.)

English1 Open the cover of battery compartment.
Pressing the upper part of the battery cover, pull the cover toward you to open.
Battery compartment

Español

Battery cover

Français* If the Blood pressure monitor is left unused for an extended period oftime, the batteries should be removed from the compartment.

2 Place batteries in the compartment.

Pyсский

Put the batteries paying attention to thepositive and negative terminal symbols
and .
4 “AA” batteries

3 Close the battery cover.

हिन्दीREMOVING THE BATTERIES• When the icon
is displayed or nothing appears in the display, replace the oldbatteries with four new ones. All four batteries should be changed at once.
• Adjust the clock after changing the batteries.
• The data stored in the memory is not deleted by changing the batteries.
• Do not use rechargeable batteries.
• Do not use alkaline (LR6) and manganese (R6P) batteries together.
• Batteries included are for demonstration purposes only. Battery life may be
shorter than specified.
• When disposing of used batteries, comply with governmental regulations orenvironmental public institution’s rules that apply in your country/area.

00_CHU304_305_2401.indb 7- Eng 7 -8/3/2024 9:21:19 AM

TIME ADJUSTMENT

START/STOP” switch
MEMORY” switch for approximately
and the “
two seconds, a beep is heard and the “month”
indicator blinks at first.

3 Press the “

MEMORY” switch to adjust the “month”.
MEMORY” switch to increase the number
Press the “
by one. Press the “ START/STOP” switch to confirm thesetting. The “month” is set and the “day” indicator blinks.
* You can fast-forward the numbers in display by pressing
MEMORY switch.
and holding the

Español

2 Press and hold both the “Françaismonitor will display time and date by turns.

Pyсский

1 When you install the batteries for the first time, theEnglishSetting the clock enables measurements to be accurately recorded.
The monitor can record the date and time of measurement as well as themeasurement results. Be sure to set the clock after inserting the batteries.

MEMORY” switch to adjust the “day”.
MEMORY” switch to increase the number
Press the “
by one. Press the “ START/STOP” switch to confirm
the setting. The “day” is set and the “hour” indicator blinks.

हिन्दी

4 Press the “

5 Press the “

MEMORY” switch to adjust the
MEMORY” switch to increase the“hour”. Press the “number by one. Press the “ START/STOP” switch toconfirm the setting. The “hour” is set and the “minutes”
indicator blinks.

6 Press the “

MEMORY” switch to adjust the
MEMORY” switch to increase“minutes”. Press the “
the number by one. Press the “ START/STOP” switch toconfirm the setting.
* Press the “ START/STOP” with the time signal on theradio, etc. to complete clock adjustment.

00_CHU304_305_2401.indb 8- Eng 8 -8/3/2024 9:21:20 AM

MEASURING YOUR BLOOD PRESSURE➊ Plug the air hose connector into
the main unit.
English• Do not bend the air hose during measurement,
which may cause inflation error, or harmful
injury due to continuous cuff pressure.

Air hoseconnector

Air hosesocket➋ ATTACHING THE CUFF

Español

Before applying your blood pressure cuff, be sure you have selected theappropriate size cuff:
Cuff (model: SCN-008) fits arm circumference: 22 cm – 30 cm.
Wide Ragne Cuff (model: SCW-009) fits arm circumference: 22 cm – 42 cm.

Français

1
2

Unroll the cuff and thread the end through themetal hoop so that the side with the hookand-loop fastener is on the outside.

Pyсский

Put the cuff around your arm so that the air
hose comes to the palm side.

Metal hoopहिन्दी
: The position of the artery.

3

Adjust the position of the cuff so that the air hose is located in thecenter of your arm and the hem of the cuff is 1 cm – 2 cm above yourelbow.

00_CHU304_305_2401.indb 9* The cuff should be put on the bare arm
or over a light-weight underwear.
* If you wear a heavy-weight top, please
remove it.
1 cm – 2 cm

- Eng 9 -8/3/2024 9:21:20 AM

EnglishPlace your arm on a table or the like,so that the center of the cuff comes tothe height of your heart.

Español

5

Pull the end of the cuff outwards sothat the cuff is snug around your arm
and then secure the end to the hookand-loop fastener.

6

Français* Carefully place the cuff on yourarm, taking care to use the correcttightness. The tightness is correct
if you can readily slide a finger
between the cuff and your arm.

Relax your arm and lightly open your hand, with the palm upward.

➌ Place the cuff at the height of the heart.
➍ Press the “START/STOP” switch.

हिन्दी

• A beep is heard and all digits displayed is an initial display, indicating that themonitor is functioning normal.
• The cuff is pressurized automatically.
* If the unit judges that pressurization is insufficient, it automatically repressurizes.
* The unit automatically sets the pressure based on the previous measurement.

Pyсский

4*	 is displayed when a pulse is detected.
*	When you want to stop measurement, press the “The cuff is deflated and measurement stops.

START/STOP” switch.

* If the previous user had a high pressure setting, that high pressure willautomatically be used for the current measurement.
* If the pressurization value exceeds 260 mmHg, if pressurization feels
abnormal, or if you want to stop the measurement process, press the“ START/STOP” switch again. The cuff deflates, and the power turns off.

00_CHU304_305_2401.indb 10- Eng 10 -8/3/2024 9:21:21 AM

➎ The measurement results are displayed.

English• Once measurement is complete, thecuff deflates and the measurement
results (systolic/diastolic blood pressure
values, pulse) are displayed.
* Systolic/diastolic blood pressure values
and pulse are displayed alternately.
* Provided there is no error in the measurement
results, they are stored automatically.

Hypertension Classification lndicator➏ Finishing measurement.
Español• Press the “ START/STOP” switch to turn the monitor off.
• After taking the blood pressure measurement, pressing the “ START/STOP”
switch will turn the monitor off. However, if you do not press the switch, themonitor’s Auto OFF function will automatically turn the monitor off after 1 minute.

Français

Irregular Heartbeat Indicator

If an irregular heartbeat is detected during measurement, the IrregularHeartbeat (IHB) Indicator is displayed after measurement is completed.

Pyсский

There are 2 conditions under which the signal of IHB will be displayed:
1) The coefficient of variation (CV) of pulse period > 25%2) T he difference of adjacent pulse period 0.14s, and the number of such
pulse takes more than 53 percentage of the total number of pulse.
• Measurement may not be correct if your heartbeat fluctuates greatly during measurement.
If the Irregular Heartbeat (IHB) Indicator is displayed, measure your blood pressure again.

CautionIf the Irregular Heartbeat (IHB) Indicator appears frequently, youare advised to consult a doctor about your health condition.

हिन्दी

Hypertension Classification Indicator
The measured blood pressure value is displayed according to the WHO standards.
See page 2)(00_CHU304_305_2401.indb 11

Systolic (mmHg)
180
160
140
130
120

Grade 3 Hypertension (severe)
Grade 2 Hypertension (moderate)
Grade 1 Hypertension (mild)
High-Normal
NormalOptimal
80

85

90 100110 DIASTOLIC(mmHg)- Eng 11 -8/3/2024 9:21:21 AMहिन्दी

■ Measurement in a reclining posture* Relax yourself to avoid placingpressure on the cuff.

EspañolYour blood pressure varies according to your posture, the time of day and a range ofother factors. Ideally, you should measure your blood pressure in the same postureat the same time every day.
■ Taking measurements while sitting*	Be seated comfortably with your feet flat on the floor, and don't
cross your legs.
➀ Place your arm on a table or similar surface with yourforearm extended.
➁ It is important to have the cuff level with your heart.
➂ Open your hand slightly on the table so that your palm
is facing up and your fingers are relaxed.
➃ Do not move your body or talk while taking themeasurement.

Français

TIPS FOR OBTAINING ACCURATE MEASUREMENTS

Pyсский

The monitor will show ‘HI’ or ‘Lo’ as technical alarm on LCD with no delay if thedetermined blood pressure (systolic or diastolic) is outside the rated range specifiedin part SPECIFICATIONS. In this case, you should consult a physician or check ifyour operation violated the instructions.
The technical alarm condition (outside the rated range) is preset in the factory andcannot be adjusted or inactivated. This alarm condition is assigned as low priority
according to IEC 60601-1-8. The technical alarm is non-latching and need no reset.
The signal displayed on LCD will disappear automatically after about 8 seconds.

EnglishTechnical Alarm Description■ You may use your right arm to measure your blood pressure.
Blood pressure value is likely to differ by as much as 10 mmHg when measured on
the right arm instead of the left (or vice-versa). Measure your blood pressure on thesame arm each day.

CALLING UP THE DATA STORED IN MEMORY
99 measurement results can be stored in
Memory.
The average value is calculated automaticallyto help you manage your daily health.

00_CHU304_305_2401.indb 12

Memory switch- Eng 12 -8/3/2024 9:21:21 AM

1

Press theMemory switch once.
Memory switch is pressed.
*	Example of display when the
Now the LCD displays the amount of the results in the current bank andindicating the average value appears at the bottom of the display. Then the
average value, based on the last three measurements, is displayed.

EnglishEspañol2

Français

If you press theMemory switch again, past measurement data aredisplayed.
E ach time you press the switch, the measurement data are displayed in order from themost recent to the oldest.
* Example of display when the
Memory switch is pressed.

Pyсский

The data number isdisplayed before themeasurement results.

हिन्दी

• The value stored in memory is numbered in the order of measurements. For example,
when 99 sets of data are stored in memory, the data number 1 represents the latest data.
Data number: 1, 2, 3, ... ... 98, 99ÇÇThe latest data
The earliest data

HOW TO DELETE THE DATA

1When any result (except average reading of the last three
Memory switch formeasurements) is displaying, press and hold themore than 3 seconds.
All data stored in Memory will be deleted.

00_CHU304_305_2401.indb 13

Press and holdfor 3 seconds
Deleted

A message saying all thedata have been deleted fromthe memory is displayed.

- Eng 13 -8/3/2024 9:21:22 AM

ABOUT “BLOOD PRESSURE”

Blood pressure value (mmHg)[Example of blood pressure variations within a day]Systolic bloodpressure

160
140
120
100
80

(Blood pressure values
measured at five-minuteintervals through the day)● Bevan AT, Honour AJ, StottFH. Clin Sci 1969;36:329–44.

60
40

Sleeping
12 13 14 15 16 17 18 19 20 21 22 23 24 1

2

3

4

5

6

7

8

Diastolic bloodpressure
9 10 11

Time of day

EspañolYour blood pressure differs according to your age, gender and a range of otherfactors. It is likely to be affected by your biorhythm during the day and by yourposture, physical activities, mental activities, level of stress and even by the airtemperature. The blood pressure of a healthy person generally varies within a day.

FrançaisYour blood pressure changes all the time.

Pyсский

(Systolic)(Diastolic)
The heart is a pump that
ArteryArterycirculates blood throughout yourLeft
Left atrium
atriumbody. Blood is pumped from theheart at a constant pressure into
arteries. This pressure is called
Left
Normal state isLeftventricle
ventricle
restored
The artery wall
the arterial blood pressure andto the artery wall.
expands.
represents, in general terms, yourDilationContraction
blood pressure. Blood pressure
is indicated by several kinds ofpressures, including the systolic pressure that occurs when the heart pumps blood,and the diastolic pressure that occurs when blood returns to the heart.

EnglishWhat is blood pressure?

Q Why is the blood pressure measured at home different from that measured by thephysician or at a hospital?
A Your blood pressure may vary due to exercise, the ambient temperature or your mental
Q
A
Q
Aहिन्दीBLOOD PRESSURE Q&A

state. When you have a physician or nurse measure your blood pressure, the pressure
value is likely to be 10 to 20 mmHg higher than usual due to anxiety and/or stress.
Knowing what affects your blood pressure helps you to better monitor your health.
Why does the blood pressure obtained vary with measurement?
Our blood pressure is adjusted by automatic nerve function. Blood pressure differswith every heart beat. We are likely to think that our blood pressure is constant, but
it varies if you take measurements in repetition. Blood pressure is susceptible to thetime of day, such as morning and afternoon, season and atmospheric temperature.
In addition, blood pressure is easily affected by mental stress or emotional ups anddowns. It tends to increase when you feel tension or decrease when you are relaxed.
What is the benefit of measuring my blood pressure at home?
Blood pressure measurements taken at home give reliable data as they can be taken
when you are in a stable and relaxed condition. Physicians place considerable importanceon blood pressure measurements taken at home. You should measure your bloodpressure at the same time each day and record the variations in your blood pressure.

Always consult your physician for an interpretation of your blood pressure
readings and to determine the proper treatment.

00_CHU304_305_2401.indb 14- Eng 14 -8/3/2024 9:21:22 AM

BEFORE REQUESTING REPAIRS OR TESTING
Make sure to check the following before sending your blood pressure monitor away
to be tested or repaired.

EnglishTroubleshooting (1)ProblemPossible Cause

Solution

Español
Français

The cuff position was not correct or Apply the cuff correctly andit was not properly tightened
try again.
Review the “BODY POSTURE
Body posture was not correct DURING MEASUREMENT”
during testing
sections of the instructions andre-test.
LCD Display
Re-test when calm and without
shows abnormal Speaking, arm or body
movement, angry, excited or
speaking or moving during
resultnervous during testing
the test.
It is inappropriate for people
Irregular heartbeatwith serious arrhythmia(arrhythmia)
to use this Electronic
Sphygmomanometer.
Troubleshooting (2)Pyсский

ProblemPossible Cause

Solution

LCD shows low
battery symbol

Low Battery

Change the batteries.

Pressure system is unstable
before measurement
LCD shows “Er 1” Fail to detect systolic pressure

LCD shows “Er 0”

Don’t move and try again.

हिन्दी

LCD shows “Er 2” Fail to detect diastolic pressure
Pneumatic system blocked or
LCD shows “Er 3” cuff
is too tight during inflation
Apply the cuff correctly andtry again.
Pneumatic system leakage or
LCD shows “Er 4” cuff
is too loose during inflation
Cuff pressure above
LCD shows “Er 5”
300 mmHg
than 3 minutes with cuff
Measure again after five
LCD shows “Er 6” Morepressure above 15 mmHg
minutes. If the monitor is still
LCD shows “Er 7”
EEPROM accessing error
abnormal, please contact the
LCD shows “Er 8” Device parameter checking error local distributor or the factory.
parameter
LCD shows “Er A” Pressure sensor
error
No response when Incorrect operation or strong
Take out batteries for five
you press button
minutes, and then reinstall allelectromagnetic
interferenceor load battery.
batteries.
* If you want to dispose this product, do not mix with general household waste. There is aseparate collection systems for used electronics products in accordance with legislation.

00_CHU304_305_2401.indb 15- Eng 15 -8/3/2024 9:21:22 AM

Weight:
Temperature
Humidity
Temperature
Storage conditions:
Humidity
Electric shock protection:
Operating conditions:

Degree of Protection

Mode of operation:
Memory:Service Life:
Cuff’s Life:
Protection against ingress of water:
Accessories:

Español
Français

Switch:Inflation:
Deflation:
Rated voltage:
Exhaust:Power supply:
Battery duration:
Automatic Power Off function:
Main Unit Dimensions:
Cuff
Cuff Size:
Wide Range Cuff

Pyсский

Iconsहिन्दी

LCD displays:

CHU304, CHU305
Oscillometric method
Digital display type
Upper Arm
Soft cuff
22 cm – 30 cm (Cuff), 22 cm – 42 cm (Wide Range Cuff)
0 – 300 mmHg
Systolic: 60 – 260 mmHg
Diastolic: 40 – 199 mmHg
40 to 180 pulse/min±3 mmHg±5% of reading
3 digits
3 digits: Measurement icon
: Ready to inflate indicator: Battery icon
: Irregular Heartbeat (IHB) Indicator: Memory icon
: Hypertension Classification Indicator
2 (START/STOP, MEMORY)
Automatic Inflation by internal pump
Automatic speed deflation system.
(: direct current)
6V DC
Electromagnetic quick exhaust valve
SIZE AA batteries (LR6)4 X 1.5V
Approx. 800 times (Alkaline)
Approx. 1 min. (after activated)
138 (W) x 54 (H) x 95 (D) mm
Approx. 148 × 510 mm
Approx. 163 × 580 mm
Unit: Approx. 207g w/o batteries
Cuff: Approx. 128g, Wide Range Cuff: Approx. 134g
10°C – 40°C
90%RH
-20°C – 55°C
90%RH
Internal power unit

EnglishSPECIFICATIONS
Model Number:
Measurement system:
Display
Measurement Localization
Cuff:
Cuff circumference range:
Cuff pressure:Pressure
Measurement range:
Pulse
Pressure
Accuracy:
Pulse
Pressure
Pulse

Type BF applied part

Continuous operation
1 × 99 readings, Average of last 3 readings
10,000 measurements or 3 years
Approx. 1000 times
IPX0
Set includes an cuff, 4 AA batteries (LR6) for themonitor, Instruction Manual
Cuff 22 cm – 30 cm, Wide Range Cuff 22 cm – 42 cm

Optional accessaries:* Applied part for this device is Cuff.
* A range in barometric pressure 800 hPa to 1050 hPa* This device corresponds to the below standards: IEC 60601-1/EN 60601-1 (Medical electricalequipment – Part 1), IEC 60601-1-2/EN 60601-1-2 (Medical electrical equipment – Part 1-2), EN
1060-3 (Non-invasive sphygmomanometers – Part 3), IEC 80601–2–30 (Medical electrical equipment– Part 2-30)

00_CHU304_305_2401.indb 16- Eng 16 -8/3/2024 9:21:22 AM

ELECTOROMAGNETIC DISTURBANCES INFORMATIONWARNING • Portable RF communications equipment should be used no closer than 30 cm (12EnglishEspañol
Français

inches) to any part of the [CHU304/305], including cables specified.
• Use of this equipment adjacent to or stacked with other equipment should be avoided.
• Use of accessories and options other than those specified (other than CITIZEN
original parts) could result in increased electromagnetic emissions or decreased
electromagnetic immunity of this equipment.
Guidance and manufacturer’s declaration - electromagnetic emissions
The [CHU304/305] is intended for use in the electromagnetic environment specified below.
The customer or the user of the [CHU304/305] should assure that it is used in such anenvironment.
Emissions test
ComplianceElectromagnetic environment - guidance
RF emissions
Group 1
The [CHU304/305] uses RF energy only for its
CISPR 11internal function. Therefore, its RF emissions
are very low and are not likely to cause any
interference in nearby electronic equipment.
RF emissions
Class BThe [CHU304/305] is suitable for use in
CISPR 11all establishments, including domestic
establishments and those directly connected toHarmonic emissions
N/A
the public low-voltage power supply network thatIEC 61000-3-2
supplies buildings used for domestic purposes.
Voltage fluctuations/ flicker N/Aemissions IEC 61000-3-3
Guidance and manufacturer’s declaration - electromagnetic immunityPyсский

The [CHU304/305] is intended for use in the electromagnetic environment specified below. The
customer or the user of the [CHU304/305] should assure that it is used in such an environment.
Immunity test

IEC 60601
test level

Electrostatic±8 kV contact
discharge (ESD)
IEC 61000-4-2 ±15 kV airहिन्दी

Electrical fasttransient/burst
IEC 61000-4-4
Surge
IEC 61000-4-5
Voltage dips,
shortinterruptionsand voltage
variations on
power supply
IEC 61000-4-11

Compliance
level

Electromagnetic environment guidance±8 kV contact

Floors should be wood, concrete or
ceramic tile. If floors are covered withsynthetic material, the relative humidity
should be at least 30%.
N/A

±15 kV air±2 kV for
N/A
power supplylines±1 kV for
input/output lines
±1 kV
N/A
line to line
±2 kV
line to earth
0% UT
0.5 cycle

N/A

N/A

N/A

30 A/mPower frequency magnetic fields should
be at levels characteristic of a typicallocation in a typical commercial or
hospital environment.

0% UT
1 cycle70% UT
25/30 cycle
0% UT
250/300 cycle

Power frequency(50/60 Hz)
30 A/mmagnetic field
IEC 61000-4-8

Note: UT is the A.C. mains voltage prior to application of the test level.

00_CHU304_305_2401.indb 17- Eng 17 -8/3/2024 9:21:23 AM

Recommended separation distances between portable and mobile RF communications
equipment and the [CHU304/305]
The [CHU304/305] is intended for use in an electromagnetic environment in which radiated RF
disturbances are controlled. The customer or the user of the [CHU304/305] can help prevent
electromagnetic interference by maintaining a minimum distance between portable and mobile
RF communications equipment (transmitters) and [CHU304/305] as recommended below,
according to the maximum output power of the communications equipment.
Rated maximum
Separation distance according to frequency of transmitter (m)output power of
150 kHz to 80 MHz
150 kHz to 80 MHz
80 MHz to 2.7 GHztransmitterd = 0.6
N/A
N/A
(W)
0.01
N/A
N/A
0.06
0.1
N/A
N/A
0.19
1
N/A
N/A
0.6
10
N/A
N/A
1.9
100
N/A
N/A
6
For transmitters rated at a maximum output power not listed above, the recommended separation distance din meters (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is
the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected byabsorption and reflection from structures, objects and people.

00_CHU304_305_2401.indb 18

EnglishEspañol
Français

10 V/m 10 V/m d = 0.6
80 MHz to 2.7 GHz
80 MHztowhere P is the maximum output power rating of the transmitter
2.7 GHz
in watts (W) according to the transmitter manufacturer and d isthe recommended separation distance in meters (m).
Field strengths from fixed RF transmitters, as determinedby an electromagnetic site survey,a) should be less than thecompliance level in each frequency range.b)Interference may occur in the vicinity ofequipment marked with the following symbol:
NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation isaffected by absorption and reflection from structures, objects and people.
a). F ield strengths from fixed transmitters, such as base stations for radio (cellular/ cordless) telephones
and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be
predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF
transmitters, an electromagnetic site survey should be considered. If the measured field strength in thelocation in which the [CHU304/305] is used exceeds the applicable RF compliance level above, the[CHU304/305] should be observed to verify normal operation. If abnormal performance is observed,
additional measures may be necessary, such as reorienting or relocating the [CHU304/305].
b). O
 ver the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

Pyсский

Radiated
RF
IEC
61000-4-3

हिन्दी

Guidance and manufacturer’s declaration - electromagnetic immunity
The [CHU304/305] is intended for use in the electromagnetic environment specified below. The customer orthe user of the [CHU304/305] should assure that it is used in such an environment.
Portable and mobile RF communications equipment should be used no closer toany part of the [CHU304/305], including cables, than the recommended separation
distance calculated from the equation applicable to the frequency of the transmitter.
Recommended separation distance
Conducted 3 Vrms, 6 Vrms N/A
N/A
150 kHz
RF
to
IEC
61000-4-6 80 MHz- Eng 18 -8/3/2024 9:21:23 AM

EnglishEspañol
FrançaisPyсский
हिन्दी

Guidance and manufacturer’s declaration - electromagnetic immunity
The [CHU304/305] is intended for use in the electromagnetic environment specified below.
The customer or the user of the [CHU304/305] should assure that it is used in such anenvironment.
Immunity to
Test
Band a) Service a)
Modulation Maximum Distance IMMUNITY
proximity fields Frequncy (MHz)
b)
power
(m)
TEST LEVEL
from RF wireless (MHz)
(W)
(V/m)
communications 385
380-390 TETRA 400
Pulse
1.8
0.3
27
equipment IEC
modulation
61000-4-3b) 18 Hz450
430-470 GMRS 460,
FM c)
2
0.3
28
FRS 460± 5 kHzdeviation
1 kHz sine
710
704-787 LTE Band
Pulse
0.2
0.3
9
13, 17
modulation
745b) 217 Hz
780
810
800-960 GSM 800/900,
Pulse
2
0.3
28
TETRA 800,
modulation
870
iDEN 820,b) 18 Hz930
CDMA 850,
LTE Band 5
1720
1700GSM 1800;
Pulse
2
0.3
28
1900
CDMA 1900;
modulation
1845
GSM 1900; DECT; b) 217 Hz
1970
LTE Band 1, 3, 4,
25; UMTS
2450
2400Bluetooth, WLAN, Pulse
2
0.3
28
2570
802.11 b/g/n,
modulation
RFID 2450,b) 217 Hz
LTE Band 7
5240
5100WLAN 802.11 a/n Pulse
0.2
0.3
9
5800
modulation
5500b) 217 Hz
5785
NOTE If necessary to achieve the IMMUNITY TEST LEVEL, the distance between thetransmitting antenna and the [CHU304/305] may be reduced to 1 m. The 1 m test distance ispermitted by IEC 61000-4-3.
a). For some services, only the uplink frequencies are included.
b). The carrier shall be modulated using a 50 % duty cycle square wave signal.
c). As an alternative to FM modulation, 50 % pulse modulation at 18 Hz may be used because
while it does not represent actual modulation, it would be worst case.
Guidance and manufacturer’s declaration - electromagnetic immunity
The [CHU304/305] is intended for use in the electromagnetic environment specified below. The
customer or the user of the [CHU304/305] should assure that it is used in such an environment.
Immunity test
Test frequency
Modulation
IMMUNITY TEST LEVEL (A/m)
Proximity magnetic
30 kHz a)CW
8b)fields134.2 kHz
Pulse modulation
65 c)
IEC 61000-4-39
2.1 kHz13.56 MHz
Pulse modulation b)
7.5 c)
50 kHza) This test is applicable only to [CHU304/305] intended for use in the HOME HEALTHCAREENVIRONMENT.
b) The carrier shall be modulated using a 50 % duty cycle square wave signal.
c) r.m.s., before modulation is applied.

00_CHU304_305_2401.indb 19- Eng 19 -8/3/2024 9:21:23 AM

00_CHU304_305_2401.indb 20

MEMO

8/3/2024 9:21:23 AM

Español
Contenido

EnglishEspañol

1 - OBSERVACIONES GENERALES
3 - PRECAUCIONES PARA EL USO Y MANTENIMIENTO
6 - IDENTIFICACIÓN DE LAS PIEZAS
7 - CÓMO REEMPLAZAR LAS PILAS
7 - CÓMO QUITAR LAS PILAS
8 - AJUSTE DE LA HORA
9 - CÓMO MEDIR LA PRESIÓN ARTERIAL
12 - CONSEJOS PARA CONSEGUIR MEDICIONES PRECISAS
12 - RECUPERACIÓN DE LOS DATOS ALMACENADOS EN LA MEMORIA
13 - ACERCA DE LA “PRESIÓN ARTERIAL”
14 - PREGUNTAS Y RESPUESTAS SOBRE LA PRESIÓN ARTERIAL
15 - ANTES DE SOLICITAR REPARACIONES O PRUEBAS
16 - ESPECIFICACIONES

Français■ Asegúrese de tener todos estos componentes.
• Monitor de presión arterial • Manual de instrucciones• 4 pilas tamaño AA (para el monitor)• Brazalete (modelo: SCN-008) para CHU304• Brazalete grande para adultos (modelo: SCW-009) para CHU305

Pyсский

Observaciones generales sobre la presión arterial y la medición dela presión arterial

OBSERVACIONES GENERALES
1.
2.

हिन्दी
3.
4.
5.
6.

Para desinflar el brazalete rápidamente, pulse el interruptor “START/STOP”.
Antes de colocar el brazalete de presión arterial, asegúrese de haberseleccionado el brazalete del tamaño correcto:
El brazalete (modelo: SCN-008) se adapta a la circunferencia del brazo:
22 cm – 30 cm.
El brazalete grande para adultos (modelo SCW-009) se adapta a una
circunferencia de brazo de: 22 cm – 42 cm.
Cuando coloque el brazalete, ajuste su posición de tal forma que la manguerade aire se encuentre en el centro del brazo y el borde entre 1 cm – 2 cm por
encima de su codo.
¡La automedición no es una tratamiento! En ningún caso deberá usted cambiar
la dosis de medicamentos recetada por su médico.
Como preparación para la medición de la presión arterial, es aconsejable queorine y después permanezca relativamente quieto durante 10 o 15 minutos
antes de la medición.
NO le dé demasiada importancia a los resultados de una sola medición. Mantenga
un registro continuado de las variaciones de la presión arterial. Solo puede
obtener un panorama completo después de una gran cantidad de lecturas.

Para obtener una correcta medición de la presión arterial
1.
2.

Respire profundamente cinco o seis veces y seguidamente relájese antes demedir la presión arterial. Si se encuentra tenso durante la medición de la presión
arterial, no conseguirá una lectura válida.
La presión arterial se elevará si se encuentra en una situación de preocupación ode enfado, de falta de sueño o estreñimiento o después del ejercicio o una comida.

00_CHU304_305_2401.indb 1- Esp 1 -8/3/2024 9:21:23 AM

Hipertensión grado 3 (grave)
Hipertensión grado 2 (moderada)
Hipertensión Grado 1 (Leve)
Alta Normal
NormalÓptimaObservaciones:
El gráfico no es exacto,
pero se puede utilizar comoorientación para comprenderlas mediciones de presión
sanguínea no invasivas.
El aparato está destinado
solamente para uso con
adultos.

Español
Français

Sistólica (mmHg)

Pyсский

Envuelva el brazalete alrededor del brazo y ajústelo de forma que pueda
Consulte la página 9, 10)
introducir un dedo entre el brazalete y el brazo. (4. NO mida la presión arterial después de bañarse o beber.
5. Mida la presión arterial en temperaturas ambiente de alrededor de 20°C. NO
mida la presión arterial cuando haga mucho frío (por debajo de 10°C) o mucho
calor (por encima de 40°C) en el lugar.
6. NO intente medir su presión arterial inmediatamente después de beber café o té o de fumar.
7. Mida la presión arterial cuando esté tranquilo y relajado. Mantenga el centro delbrazalete a la altura del corazón y NO mueva el brazo ni hable.
8. NO mida la presión arterial después de que el brazalete haya estado en su
brazo durante unos minutos. Durante este lapso, su brazo habrá desarrollado
una presión arterial mayor y no obtendrá una lectura válida.
9. Si se muestra el Indicador de latido irregular (IHB), relájese y mida de nuevo su
presión arterial. Si el indicador de latido irregular (IHB) aparece con frecuencia,
se recomienda que consulte con un médico sobre su estado de salud. (Consulte la página 11)
10. La siguiente ilustración muestra la clasificación de la presión sanguínea deacuerdo con las normas de la OMS.

English3.

हिन्दी

Diastólica (mmHg)

Definiciones y clasificaciones de los niveles de presión sanguínea* Esta unidad está equipada con un indicador que muestra visualmente laclasificación de la presión arterial (Óptima/Normal/Alta-Normal/Hipertensiónde grado 1/Hipertensión de grado 2/Hipertensión de grado 3) del resultado
después de cada medición.
11. Uso previsto• Este dispositivo es un monitor de presión arterial no invasivo mediantemétodo oscilométrico. Este dispositivo puede medir la presión arterial sistólica(SYS), la presión arterial diastólica (DIA) y el pulso de forma automática.
• Este dispositivo está diseñado para comprobar el estado de salud personalen casa, según las indicaciones de un médico, y no está pensado como
dispositivo de diagnóstico.
• Este dispositivo no está diseñado para utilizarse con recién nacidos o niños
pequeños ni tampoco para tomar mediciones cíclicas automáticas.

Mida su presión arterial a la misma hora cada día.

* La presión arterial varía a toda hora. Esto significa que los datos recogidos durante
un largo periodo tienen una mayor significación que los datos de una única
medición. Por esta razón, debe medir la presión arterial diariamente. Lo ideal es
medir su presión arterial a la misma hora cada día, siempre que sea posible.

00_CHU304_305_2401.indb 2- Esp 2 -8/3/2024 9:21:23 AM

PRECAUCIONES PARA EL USO Y MANTENIMIENTOPrecauciones para el uso
1.

EnglishEspañol
FrançaisPyсский
हिन्दी

Si sufre de enfermedades cardiaca, presión arterial alta u otra enfermedad
circulatoria, antes de utilizar el monitor, consulte con su médico.
2. El inflado a una presión más alta puede causar moretones en el lugar donde secoloca el brazalete.
Si siente que la presión del brazalete no es normal o si experimenta otra
irregularidad al utilizar el brazalete, reduzca inmediatamente la presión pulsando
el interruptor “START/STOP”, quítese el brazalete o desconecte el conector
de la manguera de aire del monitor y luego consulte con la tienda en la que
compró el monitor.
3. Si piensa que el valor medido es anormal o si la medición le hace sentirse
indispuesto, interrumpa el uso y consulte con su médico.
4. La medición de la presión arterial puede no ser posible en personas con pulso
débil o arritmia.
5. La medición repetida de la presión arterial y el inflado excesivo prolongado
pueden causar problemas como congestión o hinchazón en algunas personas.
6. Las repeticiones frecuentes de las mediciones de la presión arterial no darán
resultados precisos. Deje un intervalo de 1 minuto entre las mediciones.
7. Consulte a su médico antes de usar el monitor si usted sufre de una alteración
grave de la circulación sanguínea en los brazos. El no hacerlo puede conllevar
riesgos para la salud.
8. La medición puede no ser posible en personas con flujo sanguíneo insuficiente enel área donde se harán las mediciones o que sufre de frecuentes latidos irregulares.
Consulte con su médico para obtener consejo sobre si usar o no el monitor.
9. NO envuelva el brazalete alrededor de un brazo herido.
10. NO envuelva el brazalete alrededor de un brazo en el que haya sido colocado
un gotero (infusión intravenosa) o que esté siendo usado para transfusión
sanguínea como parte de un tratamiento médico. El hacerlo puede causar
heridas o un accidente grave.
11. No comparta el brazalete con otra persona que esté infectada, para evitar latransmisión de infecciones.
12. Si tiene alguna duda acerca de la aplicación del brazalete en el brazocorrespondiente al lado en el que se ha practicado una mastectomía, póngase
en contacto con su médico.
13. NO utilice el monitor cerca de gases inflamables como los utilizados para
anestesia. Hacer esto podría encender los gases y causar una explosión.
14. NO utilice el monitor en ambientes enriquecidos con oxígeno, como una
cámara hiperbárica de hospital o una tienda de oxígeno. Hacer esto podría
encender el oxígeno y causar un incendio.
15. NO use un teléfono móvil cerca del monitor, porque esto podría causar mal
funcionamiento.
16. Si usa un marcapasos cardiaco, consulte con su medico antes de usar elmonitor.
17. Asegúrese de utilizar esta unidad solamente para la medición de la presión
arterial. NO la utilice para ningún otro propósito.
18. Asegúrese de usar solo nuestros accesorios y piezas autorizados. Las piezas
y los accesorios no aprobados para usar con el dispositivo pueden dañar la
unidad.
19. NO utilice esta unidad en niños pequeños.
20. Es posible que no se pueda realizar la medición de la presión arterial en
personas con arritmias comunes, tales como latidos auriculares o ventricularesprematuros o fibrilación auricular.

00_CHU304_305_2401.indb 3- Esp 3 -8/3/2024 9:21:23 AM

7.
8.
9.

Advertencia

No se permite la modificación de este dispositivo.

Explicación de los símbolos: La marca CE está diseñada para proporcionar información a losinspectores de mercado de los países miembros de la Unión Europea.
: Pieza aplicada tipo BF (El brazalete es parte aplicada tipo BF).
: Mantener seco
: Advertencia: Consultar el manual de instrucciones antes del uso (Color de fondo dela señal: azul. Símbolo gráfico de la señal: blanco).
: Cumplimiento de la directiva WEEE del dispositivo: Fabricante: Símbolo de “REPRESENTANTE EN EUROPA”

Advertencia

Español5.
6.

Français

3.
4.

Pyсский

2.

NO almacene el monitor de presión arterial donde vaya a estar expuesto a laluz directa del sol, temperaturas elevadas (por encima de 55°C), temperaturas
bajas (por debajo de -20°C), humedad relativa elevada (por encima del 90%) o
en un sitio polvoriento.
Asegúrese de almacenar el monitor de presión arterial fuera del alcance deniños, mascotas y pesticidas.
NO deje caer el monitor de presión arterial o el brazalete ni los someta a otros
impactos o vibración.
Quite las pilas cuando el monitor no se vaya a utilizar durante un periodo largo.
NO intente desarmar el monitor o el brazalete. Pregunte al centro dereparaciones designado por el fabricante.
NO doble excesivamente el brazalete ni la manguera de aire.
Si el monitor y el brazalete se encuentran muy sucios, límpielos con un
paño humedecido con alcohol para esterilización o un detergente neutro. A
continuación, páseles un paño seco. En caso de ser necesario en entornos
hospitalarios, el brazalete debe desinfectarse dos veces a la semana.
NUNCA limpie el monitor de presión arterial con alcohol, disolvente o benceno,
porque esto podría dañar el monitor.
Para limpiar el brazalete, límpielo con un paño húmedo. Evite frotar con fuerza,ya que esto puede causar fugas de aire. Evite también que entre agua en la
manguera de aire.
Se recomienda limpiar el brazalete después de 200 usos.

हिन्दी

1.

EnglishPrecauciones para el mantenimiento

El equipo ha sido sometido a prueba y homologado deacuerdo con EN60601-1-2 para Dispositivos médicos
electrónicos. Esto no garantiza de ninguna forma queel dispositivo no se verá afectado por interferencia
electromagnética. Evite utilizar el dispositivo en un
entorno altamente electromagnético.

Naturaleza y frecuencia del mantenimiento:

Este producto está diseñado para su uso durante un periodo extenso;
sin embargo, generalmente se recomienda que sea inspeccionado cada dos años
para garantizar su funcionamiento y rendimiento correctos.

00_CHU304_305_2401.indb 4- Esp 4 -8/3/2024 9:21:23 AM

Protección del medioambiente:

Al deshacerse de la pila y del producto al final de su vida útil, contribuya a proteger elmedioambiente respetando las regulaciones nacionales o locales sobre reciclado.

MARCA DE WEEE
EnglishSi desea deshacerse de este producto, no lo mezcle con residuos
domésticos generales. Existe un sistema de recogida por separado deaparatos electrónicos usados, según establece la legislación de la Directiva
WEEE (Directiva 2012/19/EU) y está vigente únicamente en la Unión Europea.

Información sobre la eliminación en otros países fuera de la
Unión Europea.

Español

Si desea desechar pilas usadas, póngase en contacto con las autoridades o
distribuidor locales y pregunte cuál es el método de eliminación adecuado.

Nota sobre el símbolo de pila.

El símbolo puede aparecer en combinación con un símbolo químico. En este caso,
cumple con los requisitos establecidos por la Directiva en relación con la sustancia
química en cuestión.

Français

Estándar europeo de referencia:

El dispositivo de medición de presión arterial está en conformidad con lasregulaciones EN60601-1, EN1060-1, EN1060-3, EN1060-4.

Calibración:Pyсский

Generalmente, se recomienda inspeccionar el monitor del dispositivo de
medición de presión arterial cada dos años para garantizar su correcto
funcionamiento y precisión. Contacte con un distribuidor.

हिन्दी

Nombre:CITIZEN SYSTEMS JAPAN CO., LTD.
Dirección: 6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan

Fábrica

Nombre : ANDON HEALTH CO., LTD.
Dirección : No.3 Jinping Street, YaAn Road, Nankai District, Tianjin 300190, China

Representante europeo:

Nombre:EMERGO EUROPE
Dirección: Westervoortsedijk 60, 6827 AT Arnhem, Paises Bajos• Para obtener datos técnicos, póngase en contacto con
nuestro representante europeo.

Hecho en China

00_CHU304_305_2401.indb 5- Esp 5 -8/3/2024 9:21:23 AM

IDENTIFICACIÓN DE LAS PIEZAS
Nombres de los componentes<Unidad principal><Brazalete>

Manguera de aire

PantallaInterruptor START/STOP Stand-by

Conector para
manguera deaire

Français

Aro de metal

EspañolInterruptor
MEMORY

Enchufe de lamanguera de airePyсский

* Consulte la página 7 para obtener información del compartimiento de pilas.

<Pantalla>हिन्दी

Indicador de listo
Icono de pila para desinflar

Visualizaciónde la presión
arterial sistólica
Indicador declasificación dehipertensión

Icono dememoria
Visualización dela presión arterial
diastólica/
Visualización delas pulsaciones

Indicador de latido
irregular (IHB)*

Visualización defecha/hora* Consulte la página 11 para obtener información del indicador de latido irregular (IHB).

00_CHU304_305_2401.indb 6- Esp 6 -8/3/2024 9:21:24 AMCÓMO REEMPLAZAR LAS PILAS(Las pilas incluidas con la unidad deben introducirse en su compartimiento antes deponer el monitor de presión arterial en uso.)

English

1 Abra la tapa del compartimiento de las pilas.
Presione la parte superior de la tapa de las pilas y tire de ella hacia usted para
abrirla.
Compartimiento delas pilas

Español

Tapa delcompartimientode las pilas

Français

2 Coloque las pilas en el compartimiento.

Pyсский

3 Cierre la tapa del compartimiento de las* Si el monitor de presión arterial no se usa por un periodo extenso, laspilas se deben retirar del compartimiento.

Coloque las pilas siguiendo las indicaciones
de los símbolos de los terminales positivos
y negativosy .
4 pilas “AA”

pilas.

हिन्दी

CÓMO QUITAR LAS PILAS• Cuando se visualiza el icono
o cuando no aparece nada en la pantalla,
reemplace las pilas por cuatro nuevas. Cambie las cuatro pilas al mismo tiempo.
• Ajuste el reloj después de cambiar las pilas.
• Los datos almacenados en la memoria no se borran mediante el cambio de laspilas.
• No utilice pilas recargables.
• No utilice pilas alcalinas (LR6) y de manganeso (R6P) juntas.
• Las pilas incluidas son solo para demostración. La vida de la pila podría ser más
corta de lo especificado.
• Cuando deseche pilas usadas, cumpla con las regulaciones gubernamentales olas normas de la institución pública de medioambiente vigentes en su país/área.

00_CHU304_305_2401.indb 7- Esp 7 -8/3/2024 9:21:24 AM

AJUSTE DE LA HORA
Configurar el reloj permite que las mediciones sean registradas con
precisión.
El monitor puede registrar la fecha y la hora de las mediciones, así como losresultados de las mismas. Asegúrese de configurar el reloj después de instalar laspilas.

1 Cuando instale las pilas por primera vez, en el

MEMORY” para ajustar el
MEMORY” para aumentar“mes”. Pulse el interruptor “
el número en una unidad. Pulse el interruptor “ START/STOP” para confirmar el ajuste. El “mes” queda fijado y el
indicador del “día” comienza a parpadear.
* Puede avanzar rápidamente los números mostrados en la
MEMORY.
pantalla manteniendo pulsado el interruptor

4 Pulse el interruptor “MEMORY” para ajustar el
MEMORY” para aumentar“día”. Pulse el interruptor “
el número en una unidad. Pulse el interruptor “ START/STOP” para confirmar el ajuste. El “día” queda fijado y el
indicador de la “hora” comienza a parpadear.

Français

3 Pulse el interruptor “Pyсский

START/MEMORY” durante aproximadamente
STOP” y “
dos segundos, se escuchará un pitido y el indicadordel “mes” comenzará a parpadear.

हिन्दी

2 Mantenga pulsados los interruptores “

Españolmonitor se mostrarán la hora y la fecha de forma
secuencial.

5 Pulse el interruptor “MEMORY” para ajustar
MEMORY” parala “hora”. Pulse el interruptor “aumentar el número en una unidad. Pulse el interruptor“ START/STOP” para confirmar el ajuste. La “hora”
queda fijada y el indicador de los “minutos” comienza a
parpadear.

6 Pulse el interruptor “MEMORY” para ajustar
MEMORY” paralos “minutos”. Pulse el interruptor “aumentar el número en una unidad. Pulse el interruptor “
START/STOP” para confirmar el ajuste.
* Pulse el interruptor “ START/STOP” que presenta laseñal de hora en su radio, etc. para completar el ajuste delreloj.

00_CHU304_305_2401.indb 8- Esp 8 -8/3/2024 9:21:25 AMCÓMO MEDIR LA PRESIÓN ARTERIAL➊ Introduzca el conector para manguerade aire en la unidad principal.
English• No doble la manguera de aire durante
la medición, ya que podrían producirse
errores de inflado o lesiones provocadas
por la presión continua del brazalete.

Conector para Enchufe de lamanguera de aire manguera de aire

➋ COLOCACIÓN DEL BRAZALETE
Español
Français

12

Antes de colocar el brazalete de presión arterial, asegúrese de haberseleccionado el brazalete del tamaño correcto:
El brazalete (modelo: SCN-008) se adapta a la circunferencia del brazo:
22 cm – 30 cm.
El brazalete grande para adultos (modelo: SCW-009) se adapta a una
circunferencia de brazo de: 22 cm – 42 cm.
Desenrolle el brazalete y pase un extremo a
través del aro de metal de forma que el velcro
quede hacia afuera.

Pyсский

Colóquese el brazalete alrededor del brazode forma que la manguera de aire quede en ellado de la palma de su mano.
Aro de metalहिन्दी: la posición de la arteria.

3

Ajuste la posición del brazalete de forma que la manguera de aire
se encuentre en el centro del brazo y el borde entre 1 cm – 2 cm por
encima de su codo.

00_CHU304_305_2401.indb 9* El brazalete debe colocarse directamente
sobre la piel o también puede usarse
sobre una manga de tejido fino.
* Si usted va vestido con una prenda
gruesa, quítesela.
1 cm – 2 cm

- Esp 9 -8/3/2024 9:21:25 AM

English* Coloque el brazalete con cuidado
en su brazo, intente obtener elapriete adecuado. El apriete es
el adecuado si puede deslizar
fácilmente un dedo entre el
brazalete y el brazo.

6

Relaje el brazo y abra ligeramente la mano, con la palma hacia arriba.

➌ Coloque el brazalete a la altura del corazón.
➍ Pulse el interruptor “

START/STOP”.

• Se escucha un pitido y se muestran todos los dígitos en la pantalla inicial, lo
que indica que el monitor funciona correctamente.
• El brazalete se presuriza automáticamente.
* Si la unidad considera que la presurización es insuficiente, se vuelve a
presurizar automáticamente.
* La unidad establece la presión automáticamente basándose en la medición anterior.

Français

Español

Coloque el brazo sobre una mesao similar, de forma que el centrodel brazalete quede a la altura delcorazón.

Pyсский

5

Tire hacia afuera del extremo delbrazalete para ajustarlo alrededor delbrazo y luego asegúrelo en el extremo
con el velcro.

हिन्दी

4*	 se visualiza cuando se detecta el pulso.
*	Cuando desee detener la medición, pulse el interruptor “brazalete se desinfla y se detiene la medición.

START/STOP”. El* Si el usuario anterior tenía un ajuste de presión más alto, se utilizará esevalor de alta presión automáticamente para la medición actual.
* Si el valor de presurización excede los 260 mmHg, si la presurización
no es normal o si desea detener el proceso de medición, pulse denuevo el interruptor “ START/STOP”. El brazalete se desinfla y elequipo se apaga.

00_CHU304_305_2401.indb 10- Esp 10 -8/3/2024 9:21:25 AM

➎ Se visualizan los resultados de la medición.

English• Una vez finalizada la medición, el
brazalete se desinfla y se visualizan losresultados de la medición (valores depresión arterial sistólica/diastólica, pulso).
* Los valores de presión sanguínea
sistólica/diastólica y el pulso se
visualizan alternativamente.
* Si los resultados de la medición nocontienen errores, se almacenan
automáticamente.

lndicador de clasificación dehipertensión

Español➏ Conclusión de la medición.

Français• Pulse el interruptor “ START/STOP” para apagar el monitor.
• Después de tomar la medición de la presión arterial, pulse el interruptor“ START/STOP” para apagar el monitor. Sin embargo, si no pulsa
el interruptor, la función de apagado automático del monitor apagará
automáticamente el monitor después de 1 minuto.

Indicador de latido irregularPyсский

Si se detecta un latido irregular durante la medición, el indicador delatido irregular (IHB) se visualiza una vez que se concluya la medición.
Hay dos condiciones bajo las que se mostrará la señal de latido irregular:1) El coeficiente de variación (CV) del periodo de pulso es superior al 25 %2) La diferencia con respecto al periodo del pulso adyacente es de 0.14 s, yel número de dicho pulso supone más del 53 % del número total de pulsos.

• CPuede que la medición no sea correcta si los latidos fluctúan notablemente durante la
medición. Si se visualiza el indicador de latido irregular (IHB), mida de nuevo la presión arterial.

Si el indicador de latido irregular (IHB) aparece con
frecuencia, se recomienda que consulte con un médico
sobre su estado de salud.

Precauciónहिन्दी

Indicador de clasificación de hipertensión
El valor de presión arterial obtenido se muestra de acuerdo con las normas de la OMS.
Consulte la página 2)(
Sistólica (mmHg)

00_CHU304_305_2401.indb 11

180
160
140
130
120

Hipertensión grado 3 (grave)
Hipertensión grado 2 (moderada)
Hipertensión Grado 1 (Leve)
Alta Normal
NormalÓptima80

85

90 100110

Diastólica (mmHg)- Esp 11 -8/3/2024 9:21:26 AM

■ Medición en posición horizontal* Relájese para no ejercer presión
sobre el brazalete.
■ Puede usar el brazo derecho para medir su presión arterial.
Es probable que el valor de la presión arterial difiera en hasta 10 mmHg cuando se mide en el brazo
derecho en lugar del izquierdo (o viceversa). Mida la presión arterial en el mismo brazo todos los días.

RECUPERACIÓN DE LOS DATOS ALMACENADOS EN LA MEMORIA
Pueden almacenarse 99 resultados demedición en la memoria.
El valor promedio se calcula automáticamentepara ayudarle a administrar su salud diaria.

1

Español
Français

La presión arterial varía según su postura, la hora del día y una gama de factores. Lo
ideal es medir su presión arterial a la misma hora cada día en la misma postura.
■ Mediciones estando sentado* Siéntese cómodamente con los pies colocados en plano sobre elsuelo, y no cruce las piernas.
➀ Coloque el brazo sobre una mesa o superficie similar
con el antebrazo extendido.
➁ Es importante tener el brazalete a la altura del corazón.
➂ Abra ligeramente la mano sobre la mesa con la palma
hacia arriba y relaje los dedos.
➃ No se mueva ni hable durante el proceso de medición.

PyсскийCONSEJOS PARA CONSEGUIR MEDICIONES PRECISAS

हिन्दी

En el monitor se mostrará “HI” o “Lo” como alarma técnica en el LCD sin retraso si la presión
sanguínea determinada (sistólica o diastólica) se encuentra fuera del intervalo nominal especificado
en la sección ESPECIFICACIONES. En caso de ser así, debería ponerse en contacto con un
médico o comprobar si la operación que ha realizado supone una infracción de las instrucciones.
La condición de alarma técnica (fuera del intervalo nominal) está preajustada de fábrica, y no
puede ni ajustarse ni desactivarse. Esta condición de alarma se asigna como de baja prioridad
según la normativa IEC 60601-1-8. La alarma técnica no genera bloqueo, y no es necesario
restablecerla. La señal mostrada en la pantalla LCD desaparecerá automáticamente transcurridos
unos ocho segundos.

EnglishDescripción de la alarma técnica

Interruptor dememoria

Pulse el interruptor de memoria
una vez.
.
*	Ejemplo de pantalla cuando se pulsa el interruptor de memoria
Ahora la pantalla LCD muestra en la parte inferior la cantidad de resultadosdel banco actual y , que indica el valor promedio. Luego, se visualiza elvalor promedio, basado en las últimas tres mediciones.

00_CHU304_305_2401.indb 12- Esp 12 -8/3/2024 9:21:26 AM2

Si pulsa nuevamente el interruptor de memoria, se visualizan losdatos de las mediciones anteriores.
Cada vez que pulsa el interruptor, se visualizan los datos de medición en elorden de los más recientes a los más antiguos.
.
* Ejemplo de pantalla cuando se pulsa el interruptor de memoria

EnglishEspañol

Se visualiza el número delos datos antes que losresultados de la medición.

Français• Los valores almacenados en la memoria están enumerados según el orden
en que se tomaron las mediciones. Por ejemplo, cuando se han guardado 99
mediciones, el número 1 representa la última medición realizada.
Número de datos: 1, 2, 3, ... ... 98, 99ÇÇLa última medición La primera mediciónCÓMO BORRAR LOS DATOSPyсский

1

Mientras se muestra cualquier resultado (excepto la lectura media
de las últimas tres mediciones), mantenga pulsado el interruptor dedurante más de tres segundos.
memoria
Se borrarán todos los datos almacenados en la memoria.

हिन्दी

Mantengapulsado por 3
segundos
Borrados

Se visualiza un mensaje que
informa que todos los datos se
han borrado de la memoria.

ACERCA DE LA “PRESIÓN ARTERIAL”¿Qué es la presión arterial?

(Sistólica)(Diastólica)
El corazón es una bomba que
Arteria
Arteria
impulsa la sangre por todo el
Atrio
Atrio
izquierdo
cuerpo. La sangre es bombeadaizquierdoa presión constante desde elcorazón a las arterias. A esta
Se restaura el
Ventrículo
Ventrículo
izquierdo
izquierdo La pared arterial
estado normal en lapresión se le llama presión
pared arterial.
se expande.
sanguínea arterial y representa,
Dilatación
Contracción
en términos generales, la presión
arterial. La presión arterial está
indicada por diferentes tipos de presiones, entre ellas la presión sistólica, que ocurre
cuando el corazón bombea sangre, y la presión diastólica, que ocurre cuando lasangre regresa al corazón.

00_CHU304_305_2401.indb 13- Esp 13 -8/3/2024 9:21:27 AM

La presión arterial varía a toda hora.

120
100
80
60
40

Presión
sanguínea
diastólica

Sueño
12 13 14 15 16 17 18 19 20 21 22 23 24 1

Hora del día

2

3

4

5

6

7

8

(Valores de presión arterial
medidos en intervalos de cinco
minutos durante un día)
● Bevan AT, Honour AJ, StottFH. Clin Sci 1969;36:329–
44.

9 10 11

PREGUNTAS Y RESPUESTAS SOBRE LA PRESIÓN ARTERIALP ¿Por qué la medición de la presión arterial en casa es diferente de la tomada por elmédico o en un hospital?

R La presión arterial puede variar con el ejercicio, la temperatura ambiental o su estadode ánimo. Cuando un médico o enfermera toma su presión arterial, el valor de lapresión probablemente será 10 o 20 mmHg más alto que lo normal, debido a lapreocupación o al estrés. Saber qué es lo que afecta su presión arterial le ayudará amonitorizar mejor su salud.

P ¿Por qué los resultados de la medición de la presión arterial varían con cada

Español

140

Français

Presión
sanguínea
sistólica

160

Pyсский

Valor de la presión sanguínea(mmHg)[Ejemplo de las variaciones de la presión arterial durante un día]

EnglishLa presión arterial cambia con la edad, el sexo y una gama de otros factores. Es
probable que resulte afectada por su biorritmo durante el día y por su postura,actividades físicas, actividades intelectuales, el nivel de estrés e incluso por latemperatura atmosférica. La presión arterial de una persona saludable generalmentevaría a lo largo de un día.

difiere con cada latido del corazón. Tendemos a pensar que nuestra presión arterial
es constante, pero se puede ver cómo varía al realizar varias mediciones seguidas.
La presión arterial es diferente a diferentes horas del día, tales como la mañana y latarde, en diferentes estaciones y en diferentes temperaturas atmosféricas. Además,la presión arterial se ve fácilmente afectada por el estrés o los cambios emocionales.
Tiende a aumentar cuando se siente tensión y a disminuir cuando se está relajado.

हिन्दी

medición?

R Nuestra presión arterial es ajustada por un nervio automáticamente. La presión arterial

P ¿Cuál es el beneficio de medir la presión arterial en casa?
R La medición de la presión arterial en casa es fiable, puesto que puede tomarse en una
situación estable y relajada. Los médicos dan mucha importancia a las mediciones depresión arterial tomadas en casa. Se recomienda medir su presión arterial a la misma
hora cada día y registrar las variaciones de su presión arterial.

Siempre consulte con su médico para obtener una interpretación de laslecturas de su presión arterial y para determinar el tratamiento correcto.

00_CHU304_305_2401.indb 14- Esp 14 -8/3/2024 9:21:27 AM

ANTES DE SOLICITAR REPARACIONES O PRUEBAS
Asegúrese de comprobar los siguientes puntos antes de enviar su monitor depresión arterial a probar o reparar.
Solución de problemas (1)EnglishProblema

Español

Posible causa
Solución
La posición del brazalete no era
Coloque el brazo correctamente e
correcta o no se apretó correctamente inténtelo de nuevo.
Revise los apartados “POSICIÓN
La postura corporal durante la
CORPORAL DURANTE LA MEDICIÓN”
prueba no era la correcta
de las instrucciones y repita la prueba.
En la pantalla LCD semuestra un resultado Durante la prueba estaba hablando, Repita la prueba cuando estémovió el brazo o el cuerpo, estaba tranquilo, sin hablar ni moverse
anómalodurante la prueba.
enfadado, nervioso o alterado
Latido irregular (arritmia)

Français

El uso de este esfigmomanómetro
está contraindicado para su uso en
personas con arritmias graves.

Solución de problemas (2)Pyсский

Problema
En la pantalla LCD semuestra el símbolo depilas con poca carga
En la pantalla LCD semuestra “Er 0”
En la pantalla LCD semuestra “Er 1”
En la pantalla LCD semuestra “Er 2”

Posible causa

Solución

Pilas con poca carga

Cambie las pilas.

हिन्दी

El sistema de presión es inestableantes de la medición
No se pudo detectar la presión
No se mueva e inténtelo de nuevo.
sistólica
No se pudo detectar la presión
diastólica
El sistema neumático está
En la pantalla LCD se
bloqueado o el brazalete está
muestra “Er 3”
demasiado apretado al inflarlo
Coloque el brazalete correctamente
e inténtelo de nuevo.
El sistema neumático tiene una
En la pantalla LCD se
fuga o el brazalete está demasiado
muestra “Er 4”
holgado al inflarlo
En la pantalla LCD se La presión del brazalete está por
muestra “Er 5”
encima de los 300 mmHg
En la pantalla LCD se
Más de 3 minutos con presión delMida de nuevo dentro de cinco
muestra “Er 6”
brazalete por encima de los 15 mmHg
minutos. Si el monitor sigue en
En la pantalla LCD se
estado anormal, póngase en
Error de acceso a la EEPROM
muestra “Er 7”
contacto con el distribuidor local o
En la pantalla LCD se Error de comprobación de parámetros
con la fábrica.
muestra “Er 8”
del dispositivo
En la pantalla LCD se Error de parámetros del sensor demuestra “Er A”
presión
Extraiga las pilas durante cinco
No ha respuesta al
Funcionamiento incorrecto ominutos y, a continuación,
pulsar un botón o alinterferencia electromagnética fuerte
colóquelas de nuevo.
colocar pilas.

* Si desea deshacerse de este producto, no lo mezcle con residuos domésticos generales. Existe
un sistema de recogida por separado de aparatos electrónicos usados de acuerdo con lalegislación.

00_CHU304_305_2401.indb 15- Esp 15 -8/3/2024 9:21:27 AM

ESPECIFICACIONES
Número de modelo:
Sistema de medición:
Pantalla
Posición para medición
Brazalete:
Rango de circunferencia del brazalete:
Presión del brazalete:

Interruptor:
Inflado:Desinflado:
Voltaje nominal:
Escape:Fuente de alimentación:
Vida útil de las pilas:
Función de apagado automático:Dimensiones de la unidad principal:
BrazaleteTamaño del brazalete:
Brazalete grandepara adultos
Peso:
Temperatura
Condiciones deoperación:
HumedadTemperatura
Condiciones dealmacenamiento:
HumedadProtección contra descargas eléctricas:
Grado de protección
Modo de operación:
Memoria:
Vida útil:Vida útil del brazalete:
Protección contra la entrada de agua:
Accesorios:
Accesorios opcionales:

Español

IconosFrançais

LCD visualiza:

Pulso
Presión
Pulso
Presión
Pulso

Pyсский

Precisión

Presión

22 cm – 30 cm (brazalete), 22 cm – 42 cm (brazalete
grande para adultos)
0 – 300 mmHg
Systólica: 60 – 260 mmHg
Diastólica: 40 – 199 mmHg
40 a 180 pulsaciones/min±3 mmHg±5% de la lectura
3 dígitos
3 dígitos: icono de medición: Indicador de listo para desinflar: icono de pila
: indicador de latido irregular (IHB): icono de memoria: indicador de clasificación de hipertensión
2 (START/STOP, MEMORY)
Inflado automático mediante bomba interna
Sistema de desinflado con velocidad automática
(: corriente directa)
6V DC
Válvula electromagnética de escape rápido
tamaño AA (LR6)
4 pilas de 1.5 V
Aprox. 800 usos (alcalinas)
Aprox. 1 min (después de activado)
138 (An) × 54 (Al) × 95 (Pr) mm
Approx. 148 × 510 mm
Approx. 163 × 580 mm
Unidad: Aprox. 207g wexcluyendo pilas
Brazalete: Aprox.128g, Brazalete grande para adultos:
Aprox. 134g
10°C – 40°C
90% RH-20°C – 55°C
90% RH
Unidad de alimentación interna
Pieza aplicada tipo BF
Funcionamiento continuo
1 × 99 lecturas, promedio de las últimas 3 lecturas
10 000 mediciones o tres años
Aprox. 1000 usos
IPX0
El paquete incluye un brazalete, 4 pilas AA (LR6) para elmonitor, manual de instrucciones
Brazalete 22 cm – 30 cm, brazalete grande para adultos 22 cm – 42 cm

हिन्दी

Rango de medición:

CHU304, CHU305

Método oscilométrico
Pantalla digital
Brazo
Brazalete blando* La pieza aplicada de este dispositivo es el brazalete.
* Intervalo de presión barométrica de 800 hPa a 1050 hPa*E ste dispositivo se corresponde con los siguientes estándares: IEC 60601-1/EN 60601-1(Equipo electromédicos – Parte 1), IEC 60601-1-2/EN 60601-1-2 (Equipo electromédicos –Parte 1-2), EN 1060-3 (Esfigmomanómetros no invasivos – Parte 3), IEC 80601-2-30 (Equipos
electromédicos – Parte 2-30)

00_CHU304_305_2401.indb 16- Esp 16 -8/3/2024 9:21:27 AM

Français
Table des matières

EnglishEspañol
Français

1 - GÉNÉRALITÉS
3 - PRÉCAUTIONS D’EMPLOI ET ENTRETIEN
6 - NOMS DES PIÈCES ET LEUR EMPLACEMENT
7 - MISE EN PLACE DES PILES
7 - RETRAIT DES PILES
8 - RÉGLAGE DE L’HEURE
9 - MESURE DE LA TENSION
12 - ASTUCES POUR OBTENIR DES MESURES PRÉCISES
12 - RAPPEL DES DONNÉES MÉMORISÉES
14 - À PROPOS DE LA « TENSION ARTÉRIELLE »14 - QUESTIONS ET RÉPONSES SUR LA TENSION
15 - AVANT DE SOLLICITER UNE RÉPARATION OU DES TESTS
16 - SPÉCIFICATIONSPyсский

■ Vérifiez que toutes les pièces suivantes sont fournies.
• Tensiomètre • Mode d’emploi• 4 piles AA (pour l’appareil)• Brassard (modèle : SCN-008) pour CHU304• Grand brassard (modèle : SCW-009) pour CHU305

GÉNÉRALITÉS
Remarques générales sur la tension et sur la mesure de la tensionहिन्दी

1.
2.

3.
4.
5.
6.

Pour dégonfler rapidement le brassard, appuyez sur le bouton « START/STOP ».
Avant de fixer le brassard de tension artérielle, assurez-vous de sélectionner la taille correctede brassard.
Le brassard ordinaire (modèle : SCN-008) s’adapte à une circonférence de bras de 22 cm à 30 cm.
Le grand brassard (modèle : SCW-009) s’adapte à une circonférence de bras de 22 cm à 42 cm.
Lors de la fixation du brassard, ajustez la position du brassard de sorte que letuyau d’air soit situé au centre de votre bras et que le bord du brassard soit à
1 cm – 2 cm au-dessus de votre coude.
L’auto-mesure de la tension n’est pas une thérapie ! Vous ne devez en aucun
cas modifier la dose des médicaments prescrits par votre médecin.
En préparation de la mesure de la tension, il est recommandé d’uriner, puis derester relativement immobile pendant 10 à 15 minutes avant de réaliser la mesure.
N’accordez PAS trop d’importance aux résultats d’une seule mesure.
Consignez les écarts de tension. Un aperçu complet ne peut être obtenu qu’à
partir d’un grand nombre de lectures.

Pour obtenir une mesure correcte de la tension
1.
2.

Respirez profondément cinq à six fois puis détendez-vous avant de mesurer votre tension.
Si vous êtes tendu en réalisant cette mesure, la valeur obtenue ne sera pas valide.
La tension est élevée lorsque vous êtes dans un état d’inquiétude ou d’irritation,
si vous manquez de sommeil ou si vous êtes constipé, ou encore après une
activité physique ou un repas.

00_CHU304_305_2401.indb 1- Fra 1 -8/3/2024 9:21:27 AM

Hypertension de la catégorie 3 (grave)

Hypertension de la catégorie 2 (modérée)Remarques :Le graphique n’est pas exact,
mais il peut aider à comprendre
les mesures non invasives dela tension artérielle. L’appareil
est destiné à être utilisé sur une
population adulte uniquement.

Hypertension de la catégorie 1 (douce)
Haut-Normale
NormalOptimums

Pyсский

Français

SYSTOLIQUE(mmHg)

Español

Enroulez le brassard autour de votre bras et serrez-le de sorte que vous
puissiez y passer un doigt. (Voir page 9, 10)
4. NE mesurez PAS votre tension après un bain ou après toute consommation de liquide.
5. Mesurez votre tension dans une pièce où la température est de 20°C environ.
NE mesurez PAS votre tension lorsque la température ambiante est trop basse(inférieure à 10°C) ou trop élevée (supérieure à 40°C).
6. N’essayez PAS de mesurer votre tension immédiatement après avoir
consommé du thé ou du café ou après avoir fumé.
7. Mesurez votre tension lorsque vous êtes détendu et immobile. Maintenez le centre
du brassard au niveau de votre cœur, et NE bougez PAS votre bras ni ne parlez.
8. Ne mesurez PAS la tension alors que le brassard est en place depuis quelques minutes, carvotre bras aura accumulé une tension plus élevée, et la mesure ainsi obtenue sera incorrecte.
9. Si l’indicateur de pulsations irrégulières (IHB) apparaît, détendez-vous etmesurez à nouveau votre tension. Si l’indicateur de pulsations irrégulières (IHB)
apparaît fréquemment, nous vous recommandons de consulter un médecin au
sujet de votre état de santé.
Voir page 11)(
10. L’illustration ci-dessous présente la classification de la tension en fonction desnormes de l’OMS.

English3.

Définition et classification des niveaux de tension* Cet appareil est doté d’un indicateur qui affiche visuellement le classement
de la tension artérielle (optimums/normal/haut-normale/hypertension de lacatégorie 1/hypertension de la catégorie 2/hypertension de la catégorie 3) durésultat après chaque mesure.
11. Utilisation prévue• Ce dispositif est un tensiomètre non invasif par méthode oscillométrique. Il
permet de mesurer automatiquement la pression artérielle systolique (SYS), lapression artérielle diastolique (DIA) et la fréquence du pouls.
• Ce dispositif est destiné à être utilisé pour vérifier son état de santé chez soi
conformément aux directives d’un médecin et n’est pas conçu comme unappareil de diagnostic.
• Ce dispositif n’est pas destiné à être utilisé avec un nouveau-né ou un
nourrisson, et n’est pas conçu pour la mesure automatique de cycles.

हिन्दी

DIASTOLIQUE(mmHg)

Mesurez votre tension chaque jour à la même heure.

* Votre tension change en permanence. Cela signifie que les données recueillies surune longue période ont une plus grande signification que celles issues d’une seulemesure. Il est donc nécessaire de mesurer votre tension quotidiennement. Dansl’idéal, vous devriez mesurer votre tension chaque jour à la même heure si possible.

00_CHU304_305_2401.indb 2- Fra 2 -8/3/2024 9:21:27 AM

PRÉCAUTIONS D’EMPLOI ET ENTRETIEN
Précautions d’emploi
1.

EnglishEspañol
FrançaisPyсский
हिन्दी

Si vous souffrez d’une maladie cardiaque, si vous faites de l’hypertension ou
si vous souffrez de toute autre maladie de la circulation, veuillez consulter unmédecin avant d’utiliser cet appareil.
2. Gonfler le brassard à une pression plus élevée peut entraîner des contusions à
l’endroit où le brassard a été mis en place.
Si, pendant une prise de mesure, vous ressentez une pression anormale au
niveau du brassard, ou si vous constatez une quelconque irrégularité, diminuez
la pression en appuyant immédiatement sur le bouton « START/STOP » etretirez le brassard ou débranchez le connecteur du tuyau d’air de l’appareil.
Prenez ensuite contact avec le point de vente où vous avez acheté l’appareil.
3. Si vous pensez que la mesure est anormale ou si le fait de mesurer votre tension
provoque chez vous un malaise, arrêtez la mesure et consultez votre médecin.
4. La mesure de la tension peut s’avérer impossible pour une personne ayant unpouls faible ou souffrant d’arythmie.
5. En cas de mesures répétées de la tension ou de surgonflage prolongé, unecongestion ou un gonflement peut se produire chez certaines personnes.
6. En cas de mesures répétées de la tension, aucun résultat précis ne sera
obtenu. Un intervalle d’une minute entre deux mesures doit être respecté.
7. Consultez votre médecin avant d’utiliser cet appareil si vous souffrez d’un grave
trouble de la circulation sanguine dans les bras. Sinon, cela pourrait se révéler
dangereux pour votre santé.
8. La mesure peut ne pas être possible pour une personne ayant un flux sanguin
insuffisant dans la zone de mesure ou des battements de cœur irréguliers.
Consultez votre médecin pour obtenir son avis sur l’utilisation de cet appareil.
9. NE placez PAS le brassard autour d’un bras blessé.
10. NE placez PAS le brassard sur un bras dans lequel une perfusion a été insérée(injection intraveineuse) ou s’il est utilisé pour une transfusion sanguine dansle cadre d’un traitement médical. Le non-respect de cette consigne risque deprovoquer une blessure ou un accident grave.
11. Ne prêtez pas le brassard à une personne souffrant d’une infection pour éviter
toute contamination.
12. Si vous devez installer le brassard sur le côté d’une masectomie, n’hésitez pasà demander conseil à un médecin.
13. N’utilisez PAS cet appareil à proximité de gaz inflammables comme les gaz
anesthésiants en raison du risque d’incendie et d’explosion.
14. N’utilisez PAS cet appareil dans des environnements enrichis en oxygène
comme la chambre hyperbare d’un hôpital ou une tente à oxygène en raison durisque d’incendie.
15. N’utilisez PAS de téléphone portable à proximité de cet appareil sous peine deprovoquer un dysfonctionnement.
16. Si vous utilisez un stimulateur cardiaque, veuillez consulter votre médecin avantd’utiliser cet appareil.
17. N’utilisez cet appareil que pour mesurer la tension. NE l’utilisez PAS à d’autres
fins.
18. Veillez à n’utiliser que nos pièces et accessoires autorisés. Les pièces etaccessoires dont l’utilisation n’est pas approuvée avec le dispositif peuvent
l’endommager.
19. N’utilisez PAS cet appareil sur un nourrisson.
20. La mesure de la tension artérielle peut s’avérer impossible pour des personnes
souffrant d’arythmies courantes comme une extrasystole atriale ou ventriculaire,
ou une fibrillation atriale.

00_CHU304_305_2401.indb 3- Fra 3 -8/3/2024 9:21:27 AM

7.
8.
9.

Avertissement

Aucune modification de ce dispositif n’est autorisée.

Explication des symboles: Le marquage CE est destiné à fournir des informations aux
inspecteurs de marché dans les pays membres de l’UE.
: Pièce de type BF appliquée sur le patient (le brassard est une pièce detype BF appliquée sur le patient).
: Garder au sec: Avertissement: Consulter le mode d’emploi avant utilisation (Couleur du fond del’indication : bleu. Symbole graphique de l’indication : blanc).
: Directive DEEE sur la conformité des appareils électriques: Fabricant: Symbole pour la « REPRÉSENTATION EUROPÉENNE »Avertissement

Español5.
6.

Français

3.
4.

Pyсский

2.

NE conservez PAS le tensiomètre dans un endroit exposé à la lumière
directe du soleil, à des températures élevées (supérieures à 55°C), à destempératures basses (inférieures à -20°C), à un taux d’humidité relativementélevé (supérieure à 90%) ou dans un endroit très poussiéreux.
Assurez-vous de ranger le tensiomètre hors de la portée des enfants et desanimaux domestiques.
NE faites PAS tomber le tensiomètre ni le brassard et ne le soumettez à aucune
force ni vibration.
Retirez les piles pendant les périodes d’inutilisation prolongée.
NE démontez PAS le tensiomètre ni le brassard. Adressez-vous au centre deréparation désigné par le fabricant.
NE pliez PAS le brassard ni le tuyau d’air de façon excessive.
Si le tensiomètre et le brassard sont très sales, nettoyez-les avec un chiffon
humide imprégné d’alcool ou d’un détergent neutre, puis essuyez-les avec un
chiffon sec. Dans le cadre d’un usage en milieu hospitalier, le brassard doit êtredésinfecté deux fois par semaine.
Ne nettoyez JAMAIS le tensiomètre avec de l’alcool, du diluant ou du benzène
sous peine de le détériorer.
Nettoyez le brassard avec un chiffon humide. Évitez de frotter trop fort souspeine de provoquer des fuites d’air. Prenez également soin de ne pas laisserd’eau s’infiltrer dans le tuyau d’air.
Nous recommandons un nettoyage du brassard toutes les 200 utilisations.

हिन्दी

1.

EnglishPrécautions d’entretien

L'appareil a été testé et homologué conformément à lanorme EN60601-1-2 pour les appareils électromédicaux.
Ceci ne constitue en aucun cas une garantie que cetappareil ne sera pas affecté par des interférencesélectromagnétiques. Évitez d’utiliser cet appareil dans unenvironnement hautement électromagnétique.

Nature et fréquence de l’entretien :Cet appareil a été conçu pour une utilisation pendant une période prolongée ;il est toutefois généralement recommandé de l’inspecter tous les deux ans afin devérifier son bon fonctionnement et ses performances.

00_CHU304_305_2401.indb 4- Fra 4 -8/3/2024 9:21:28 AM

Protection de l’environnement :Merci d’aider à protéger l’environnement en respectant les réglementations nationales
et/ou locales de recyclage lorsque vous mettez au rebut la pile et le produit en fin de vie.

MARQUAGE DEEE
EnglishSi vous souhaitez vous débarrasser de cet appareil, ne le mettez pas à lapoubelle avec vos ordures ménagères. II existe un système de récupération
distinct pour les appareils électroniques usagés conformément à la législation
dans le cadre de la Directive DEEE (Directive 2012/19/EU) qui est uniquement
valable dans les pays de l’Union européenne.

Español

Si vous souhaitez mettre au rebut les piles usées, veuillez contacter les autorités localesou le revendeur pour vous renseigner sur la méthode appropriée de mise au rebut.

Français

Informations sur la mise au rebut dans les autres pays en dehors del’Union européenne.

Norme de référence européenne :Remarque sur le symbole des piles.

Ce symbole peut être utilisé en association avec un symbole chimique. Dans ce cas,il est conforme aux prescriptions mises en place par la Directive relative aux produits
chimiques concernés.
Les appareils de mesure de la pression artérielle correspondent aux réglementations
EN60601-1, EN1060-1, EN1060-3, EN1060-4.

Calibrage :Pyсский

Il est généralement recommandé d’inspecter l’appareil de mesure de la pression
artérielle tous les deux ans afin de vérifier son bon fonctionnement et sa précision.
Veuillez contacter un distributeur.

हिन्दी

Nom :Adresse :

Usine

Nom :
Adresse :CITIZEN SYSTEMS JAPAN CO., LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan
ANDON HEALTH CO., LTD.
No.3 Jinping Street, YaAn Road, Nankai District, Tianjin 300190, China

Représentant en Europe:
Nom :Adresse :EMERGO EUROPE
Westervoortsedijk 60, 6827 AT Arnhem, Pays-Bas• Pour les données techniques, veuillez contacter notre
représentant en Europe.

Fabriqué en Chine

00_CHU304_305_2401.indb 5- Fra 5 -8/3/2024 9:21:28 AM

NOMES DES PIÈCES ET LEUR EMPLACEMENT
Noms des pièces
Tuyau d’air

English<Brassard>

Affichage

BoutonMEMORY

Connecteur dutuyau d’air

Français

Bouton START/STOP Veille

Crochetmétallique

Español<Unité principale>Branchement dutuyau d’air<Affichage>
Indicateur Prêt à
Icône de pile gonfler

Affichage dela pression
artérielle
systolique
Indicateur declassificationd’hypertensionहिन्दी

Pyсский

* Voir page 7 pour le compartiment à piles.

Icône demémoireAffichage dela pression
artérielle
diastolique/dupouls

Indicateur depulsations irrégulières(IHB)*

Affichage de la date/heure* Voir page 11 pour l’indicateur de pulsations irrégulières (IHB).

00_CHU304_305_2401.indb 6- Fra 6 -8/3/2024 9:21:28 AM

MISE EN PLACE DES PILES(Les piles fournies avec l’appareil doivent être chargées dans le compartiment avant
l’utilisation du tensiomètre.)

English1 Ouvrez le couvercle du compartiment à piles.
En appuyant sur la partie supérieure du couvercle du compartiment à piles, tirez
le couvercle vers vous pour l’ouvrir.
Compartiment à piles

Español

Couvercle ducompartimentà pilesFrançais* Si vous prévoyez de ne pas utiliser le tensiomètre pendant une
période prolongée, retirez les piles du compartiment.

2 Placez les piles dans le compartiment.

Pyсский

Insérez les piles en respectant le sens despolarités plus et moins .
4 piles (format AA)

3 Fermez le couvercle du compartiment à
piles.

हिन्दीRRETRAIT DES PILES• Lorsque l’icône
apparaît ou si rien ne s’affiche à l’écran, remplacez lespiles usées par quatre piles neuves. Il est important de changer les piles
simultanément.
• Réglez l’horloge après avoir remplacé les piles.
• Les données se trouvant dans la mémoire ne sont pas effacées lors duremplacement des piles.
• N’utilisez pas de piles rechargeables.
• N’utilisez pas ensemble des piles alcalines (LR6) et des piles au manganèse (R6P).
• Les piles fournies le sont uniquement dans un but de démonstration. La durée devie des piles peut être plus courte que la durée indiquée.
• Lorsque vous jetez des piles usées, veuillez vous plier aux règlements nationaux ou
aux règles en matière d’environnement en vigueur dans votre pays ou votre région.

00_CHU304_305_2401.indb 7- Fra 7 -8/3/2024 9:21:28 AM

RÉGLAGE DE L’HEURE

3 Appuyez sur le bouton «MEMORY » pour régler
le « mois ». Appuyez sur le bouton «MEMORY » pouraugmenter le nombre d’une unité. Appuyez sur le bouton« START/STOP » pour valider le réglage. Le « mois » est
réglé et l’indicateur « jour » clignote..
* Vous pouvez faire avancer rapidement les nombres surl’affichage en maintenant enfoncé le bouton
MEMORY.

4 Appuyez sur le bouton «MEMORY » pour régler
le « jour ». Appuyez sur le bouton «MEMORY » pouraugmenter le nombre d’une unité. Appuyez sur le bouton« START/STOP » pour valider le réglage. Le « jour » est
réglé et l’indicateur « heure » clignote.

Español

START/STOP » et le bouton «MEMORY » pendant deux
secondes environ. Un bip est émis et l’indicateur «mois » se met à clignoter.

Français

2 Maintenez enfoncés à la fois le bouton «Pyсский

première fois, l’appareil affiche successivement
l’heure et la date.

हिन्दी

1 Lorsque vous mettez en place les piles pour laEnglishRégler l’horloge permet aux mesures d’être enregistrées avec précision.
L’appareil peut enregistrer la date et l’heure d’une mesure ainsi que les résultats dela mesure. Assurez-vous de régler l’horloge après avoir mis en place les piles.

5 Appuyez sur le bouton «MEMORY » pour régler
l’« heure ». Appuyez sur le bouton «MEMORY » pouraugmenter le nombre d’une unité. Appuyez sur le bouton« START/STOP » pour valider le réglage. L’« heure » est
réglée et l’indicateur « minutes » clignote.

6 Appuyez sur le bouton «MEMORY » pour régler
les « minutes ». Appuyez sur le bouton «MEMORY »pour augmenter le nombre d’une unité. Appuyez sur lebouton « START/STOP » pour valider le réglage.
* Appuyez sur le bouton « START/STOP » en accord
avec le signal horaire de la radio ou autre pour valider leréglage de l’horloge.

00_CHU304_305_2401.indb 8- Fra 8 -8/3/2024 9:21:30 AM

MESURE DE LA TENSION➊ Branchez le connecteur du tuyau
d’air dans l’unité principale.
English• Ne pliez pas le tuyau d’air pendant la mesure
pour éviter toute erreur de gonflage ou blessure Connecteur du Branchementtuyau d’air du tuyau d’air
sous l’effet de la pression du brassard.

➋ FIXATION DU BRASSARD

Español

Avant de fixer le brassard de tension artérielle, assurez-vous de sélectionnerla taille correcte de brassard.
Le brassard ordinaire (modèle : SCN-008) s’adapte à une circonférence debras de 22 cm à 30 cm.
Le grand brassard (modèle : SCW-009) s’adapte à une circonférence debras de 22 cm à 42 cm.

Français1

Pyсский

2

Déroulez le brassard et faites passer l’extrémité
par le crochet métallique de sorte que lecôté portant la bande velcro soit tourné vers
l’extérieur.
Mettez le brassard autour de votre bras desorte que le tuyau d’air se trouve du côté devotre paume.
Crochetmétalliqueहिन्दी
: L’emplacement de l’artère.

3

Ajustez la position du brassard de sorte que le tuyau d’air soit situé aucentre de votre bras et que le bord du brassard soit à 1 cm – 2 cm
au-dessus de votre coude.
* Le brassard doit être placé directement
sur le bras nu, mais il peut égalementêtre utilisé par dessus la manche d’unvêtement léger.
* Si vous portez un haut épais, retirez-le.

00_CHU304_305_2401.indb 9

1 cm – 2 cm

- Fra 9 -8/3/2024 9:21:30 AM

EnglishPlacez votre bras sur une table ou
similaire de manière à ce que le centre dubrassard soit au niveau de votre cœur.

Español5

Tirez sur l’extrémité du brassard
afin de l’ajuster autour du bras, puis
attachez l’extrémité à l’aide de labande velcro.

* Positionnez le brassard sur lebras avec précaution en veillant
à serrer correctement. Vous avez
serré correctement si vous pouvez
facilement glisser un doigt entre lebrassard et votre bras.

6

Détendez votre bras et ouvrez légèrement la paume de la main vers lehaut.

➍ Appuyez sur le bouton «Pyсский

➌ Placez le brassard à la hauteur de votre cœur.
START/STOP ».

हिन्दी

• Un bip est émis et tous les chiffres s’affichent au début indiquant quel’appareil fonctionne normalement.
• Le brassard est automatiquement gonflé.
* Si l’appareil juge que la pression est insuffisante, il regonfle
automatiquement le brassard.
* L’appareil règle automatiquement la pression d’après la mesure
précédente.

*	 s’affiche lorsqu’une pulsation est détectée.
*	Lorsque vous souhaitez arrêter la mesure, appuyez sur le bouton «STOP ». Le brassard se dégonfle et la mesure s’arrête.

Français

4

START/* Si l’utilisateur précédent a défini un réglage de pression élevée, cettepression élevée sera automatiquement utilisée pour la mesure actuelle.
* Si la valeur de la pression est supérieure à 260 mmHg, si elle semble
anormale ou si vous souhaitez interrompre la procédure de mesure,
appuyez une nouvelle fois sur le bouton « START/STOP ». Le
brassard se dégonfle et l’appareil s’éteint.

00_CHU304_305_2401.indb 10- Fra 10 -8/3/2024 9:21:30 AM

➎ Les résultats de la mesure s’affichent.

English• Une fois que la mesure est terminée, lebrassard se dégonfle et les résultats dela mesure (pression artérielle systolique/diastolique, pouls) s’affichent.
* Les valeurs de la tension systolique/diastoliqueet le pouls sont affichés tour à tour.
* Ces valeurs sont automatiquement
mémorisées sous réserve que les
Indicateur de classification d’hypertension
résultats de la mesure ne présentent
aucune erreur.

Español➏ Terminez la mesure.
• Appuyez sur le bouton «START/STOP » pour éteindre l’appareil.
• Une fois la tension artérielle mesurée, appuyez sur le bouton « START/STOP » pouréteindre l’appareil. Même si vous n’appuyez pas sur le bouton, la fonction de mise horstension automatique éteindra automatiquement l’appareil au bout d’une minute.

Français

Indicateur de pulsations irrégulières

Si des battements de cœur irréguliers sont détectés pendant la mesure,l’indicateur de pulsations irrégulières (IHB) apparaît une fois la mesure terminée.

Pyсский

Le signal de IHB s’affichera dans deux cas de figure :1) Coefficient de variation (CV) de la période de pulsation > 25 %2) D ifférence de période de pulsation adjacente 0,14 sec. et le nombre deces pulsations représente plus de 53 % du nombre total de pulsations.
• La mesure peut ne pas être correcte si vos battements de cœur fluctuent
considérablement pendant la mesure. Si l’indicateur de pulsations irrégulières (IHB)
apparaît, mesurez à nouveau votre tension artérielle.

Attentionहिन्दी

Si l’indicateur de pulsations irrégulières (IHB) apparaît
fréquemment, nous vous recommandons de consulter un médecin
au sujet de votre état de santé.

Indicateur de classification d’hypertension
La valeur de la tension artérielle mesurée s’affiche d’après les normes de l’OMC.
Voir page 2)(00_CHU304_305_2401.indb 11

SYSTOLIQUE (mmHg)
180
160
140
130
120

Hypertension de la catégorie 3 (grave)
Hypertension de la catégorie 2 (modérée)Hypertension de la catégorie 1 (douce)
Haut-Normale
NormalOptimums
80

85

90 100- Fra 11 -110

DIASTOLIQUE(mmHg)

8/3/2024 9:21:30 AM

■ Prendre la mesure en position assise*	Asseyez-vous confortablement, les pieds posés à plat sur le sol,jambes décroisées.
➀ Placez le bras sur une table ou sur une surface
semblable avec l’avant-bras étiré.
➁ Il est important de positionner le brassard au niveau
de votre cœur.
➂ Ouvrez légèrement la main sur la table de sorte que
votre paume soit tournée vers le haut et que vos
doigts soient détendus.
➃ Ne bougez pas le corps et ne parlez pas pendant lamesure.

Españolहिन्दी

■ Mesure en position allongée* Détendez-vous pour ne pas
exercer de pression sur lebrassard.

Français

ASTUCES POUR OBTENIR DES MESURES PRÉCISES
Votre tension varie en fonction de votre posture, de l’heure de la journée et d’unéventail d’autres facteurs. Dans l’idéal, vous devriez mesurer votre tension dans lamême posture chaque jour à la même heure.

Pyсский

Le message ‘HI’ ou ‘Lo’ s’affiche immédiatement sur l’écran LCD du tensiomètre si
la pression artérielle mesurée (systolique ou diastolique) est en-dehors de la plage
de mesure spécifiée au paragraphe SPÉCIFICATIONS. Dans ce cas, consultez unmédecin et vérifiez que vous utilisez l’appareil conformément aux instructions.
L’alarme technique (hors plage de mesure) est paramétrée par défaut et ne peut êtreajustée ou désactivée. Cette alarme est un dispositif de faible priorité conformément
à la norme IEC 60601-1-8. L’alarme technique n’est pas verrouillable et ne nécessite
pas de réinitialisation. Le signal affiché sur l’écran LCD disparaît automatiquement aubout de 8 secondes.

EnglishDescription de l’alarme technique■ Vous pouvez utiliser votre bras droit pour mesurer votre tension
artérielle.
La valeur de la tension artérielle peut différer de 10 mmHg lorsqu’elle est mesurée
sur le bras gauche au lieu du droit (ou vice et versa). Mesurez votre tension sur lemême bras chaque jour.

RAPPEL DES DONNÉES MÉMORISÉES
99 résultats de mesures peuvent êtremémorisés.
La valeur moyenne est calculéeautomatiquement pour vous aider à gérer votre
santé quotidienne.

00_CHU304_305_2401.indb 12

BoutonMémoire- Fra 12 -8/3/2024 9:21:31 AM

1Appuyez une fois sur le bouton Mémoire.
.
*	Exemple d’affichage lors de la pression sur le bouton MémoireÀ présent, l’écran LCD affiche la somme des résultats dans le stock actuel, et
indiquant la valeur moyenne apparaît en bas de l’affichage. Puis, la valeur
moyenne basée sur les trois dernières mesures est affichée.

EnglishEspañol2

Français

Si vous appuyez de nouveau sur le bouton Mémoire, les données
passées sont affichées.
Chaque fois que vous appuyez sur le bouton, les données des mesures sontrappelées dans l’ordre des plus récentes aux plus anciennes.
* Exemple d’affichage lors de la pression sur le bouton Mémoire.

Pyсский

Le numéro des données
est affiché avant lesrésultats de la mesure.

हिन्दी

• Les valeurs mémorisées sont numérotées dans l’ordre des mesures. Par exemple
quand 99 ensembles de données sont mémorisés, les données avec le numéro 1
représentent les données les plus récentes.
Numéro de données: 1, 2, 3, ...
... 98, 99
ÇÇDonnées les plus Données les plus
récentes
anciennes

COMMENT EFFACER LES DONNÉES

1

Pendant l’affichage des résultats (sauf le relevé moyen des trois
dernières mesures), maintenez enfoncé le bouton
Memory pendant
plus de 3 secondes.
Toutes les données mémorisées sont effacées.

00_CHU304_305_2401.indb 13

Maintenez
enfoncé pendant
3 secondes
Effacé

Un message indiquant que
toutes les données ont étéeffacées de la mémoire apparaît.

- Fra 13 -8/3/2024 9:21:31 AM

À PROPOS DE LA « TENSION ARTÉRIELLE »Qu’est-ce que la tension artérielle ?

Ventricule
Les parois de l’artère
gauches’agrandissent.
Contraction

Ventricule
gaucheArtèreRetour à l’état
normal de la paroi
de l’artère
Dilatation

Votre tension change en permanence.

La tension varie en fonction de l’âge, du sexe et d’un éventail d’autres facteurs. La
tension d’une personne est susceptible d’être influencée par le biorythme d’une journée,par la position, par l’activité physique, par l’activité mentale, par le niveau de stress etpar la température de l’air. La tension artérielle d’une personne en bonne santé variegénéralement au cours d’une journée.
Pression artérielle
systolique

160
140
120
100
80
60
40

Sommeil
12 13 14 15 16 17 18 19 20 21 22 23 24 1

2

3

4

5

Moment de la journée

6

7

8

(Valeurs de la tension artérielle
mesurées à des intervalles decinq minutes sur une journée)● Bevan AT, Honour AJ, StottFH. Clin Sci 1969;36:329–44.

Pyсский

Valeur de tension (mmHg)[Exemple de variations de tension artérielle sur une journée]Pression artérielle
diastolique
9 10 11QUESTIONS ET RÉPONSES SUR LA TENSIONQ Pourquoi est-ce que la tension mesurée à la maison est différente de celle mesuréedans le cabinet d’un médecin ou à l’hôpital ?
R Votre tension est susceptible de varier en fonction de l’activité physique, de la
Q
R

Q
R

EnglishAtriumgaucheEspañol(Diastolique)ArtèreAtriumgaucheFrançais(Systolique)température ambiante ou de votre état psychologique. Lorsqu’un médecin ou une
infirmière prend votre tension, la valeur de la tension est probablement supérieure de
10 à 20 mmHg à la valeur habituelle en raison de votre anxiété et/ou stress. Connaître
ce qui influence votre tension vous aidera à mieux surveiller votre santé.
Pourquoi la tension artérielle obtenue varie-t-elle à chaque mesure ?
Notre tension artérielle est réglée par la fonction d’un nerf automatique. Elle varieà chaque battement de cœur. On a tendance à penser que notre tension est
constante, mais vous verrez qu’elle change si vous prenez plusieurs mesures à lasuite. La tension artérielle est sensible au moment de la journée, comme le matin etl’après-midi, la saison et la température atmosphérique. De plus, la tension artérielle
est facilement influencée par le stress mental et les émotions. Elle a tendance à
augmenter lorsque vous vous sentez tendu et à baisser lorsque vous êtes détendu.
Quel est l’avantage de mesurer ma tension à domicile ?
La tension mesurée à domicile offre des données fiables dans la mesure où elle peutêtre réalisée dans un état stable et détendu. Les médecins accordent une grande
importance aux mesures de tension réalisées à domicile. Vous devriez mesurer votre
tension chaque jour à la même heure et consigner les écarts de tension.

हिन्दी

Le cœur est une pompe qui assure la circulation
du sang dans le corps. Le sang est pompé àpartir du cœur à pression constante en direction
des artères. Cette pression est appelée tension
artérielle et représente, en général, votre tension.
La tension est indiquée par divers types depression comme : la pression systolique qui se
produit lorsque le cœur pompe le sang, et lapression diastolique qui se produit lorsque lesang retourne vers le cœur.

Consultez toujours votre médecin pour l’interprétation de vos mesures detension et pour déterminer le traitement adapté.

00_CHU304_305_2401.indb 14- Fra 14 -8/3/2024 9:21:31 AM

AVANT DE SOLLICITER UNE RÉPARATION OU DES TESTS
Vérifiez les points ci-dessous avant de faire tester ou réparer votre tensiomètre.
Dépannage (1)EnglishProblème

EspañolL’affichage LCD
indique un résultat
anormal

Cause possible
Le brassard était mal positionné ou
incorrectement serré
La position corporelle était
incorrecte pendant la mesure
Conversation, mouvement du brasou du corps, colère, excitation ou
nervosité pendant la mesure
Pulsations irrégulières (arythmie)

Solution
Placez correctement le brassard etréessayez.
Consultez les sections « POSITIONDU CORPS PENDANT LA
MESURE » des instructions etrecommencez la mesure.
Recommencez la mesure une fois
que vous êtes calme et sans parler
ni bouger.
Les personnes souffrant d’unearythmie grave ne doivent pas
utiliser ce tensiomètre électronique.

Français

Dépannage (2)Pyсский

Problème
L’affichage LCD
indique un symbolede pile faible
L’affichage LCD
indique « Er 0 »L’affichage LCD
indique « Er 1 »L’affichage LCD
indique « Er 2 »L’affichage LCD
indique « Er 3 »हिन्दी

L’affichage LCD
indique « Er 4 »L’affichage LCD
indique « Er 5 »L’affichage LCD
indique « Er 6 »L’affichage LCD
indique « Er 7 »L’affichage LCD
indique « Er 8 »L’affichage LCD
indique « Er A »Pas de réponseen appuyant surles boutons ou en
insérant la pile.

Cause possible

Solution

Pile faible

Remplacez les piles.

Le système de pression est instableavant la mesure
La pression systolique n’est pas
détectée
La pression diastolique n’est pas
détectée
Le système pneumatique est
bloqué ou le brassard est trop serré
pendant le gonflage
Le système pneumatique fuit ou
le brassard n’est pas assez serré
pendant le gonflage
Pression du brassard au-dessus de
300 mmHg
Plus de 3 minutes avec la pression
du brassard au-dessus de 15
mmHg
Erreur d’accès EEPROM
Erreur de vérification des paramètres
de l’appareil
Erreur des paramètres du capteurde pression
Fonctionnement incorrect ouinterférence électromagnétiqueimportante

Ne bougez pas et réessayez.

Placez correctement le brassard etréessayez.

Mesurez à nouveau après cinq
minutes. Si l’appareil présente
toujours une anomalie, veuillez
contacter le distributeur local ou lefabricant.

Retirez les piles pendant cinq
minutes, puis remettez-les en place.

* Si vous souhaitez vous débarrasser de cet appareil, ne le mettez pas à la poubelle avec vos
ordures ménagères. Il existe un système de récupération distinct pour les appareils électroniques
usagés conformément à la législation.

00_CHU304_305_2401.indb 15- Fra 15 -8/3/2024 9:21:31 AM

00_CHU304_305_2401.indb 16

Español
FrançaisPyсский
हिन्दी

CHU304, CHU305
Méthode oscillométrique
Affichage numériquePartie supérieure du bras
Brassard souple
22 cm – 30 cm (Brassard), 22 cm – 42 cm (Grand brassard)
0 – 300 mmHg
Systolique : 60 – 260 mmHg
Pression
Diastolique : 40 – 199 mmHg
Plage de mesure :Pouls
40 à 180 pulsations/minutePression±3 mmHg
Précision :Pouls
à ±5% près
Pression
3 chiffres
Pouls
3 chiffres: Icône de mesure: Indicateur Prêt à gonfler
Écrans LCD :: Icône de pile
Icônes: Indicateur de pulsations irrégulières (IHB): Icône de mémoire: Indicateur de classification d’hypertension
Bouton :2 (START/STOP, MEMORY)
Gonflement :Gonflement automatique par pompe interne
Dégonflement :Système de dégonflement automatique(: courant direct)
Tension nominale :6 VCC
Échappement d’air :Soupape électromagnétique d’évacuation rapide(LR6)
Alimentation :4 X piles AA 1.5 V
Durée des piles :Environ 800 utilisations (Alcaline)
Fonction de mise hors tension automatique :Environ 1 minutes (après avoir été activé)
Dimensions de l’unité principale :138 (L) x 54 (H) x 95 (P) mm
Brassard
Environ 148 x 510 mm
Taille du brassard :Grand brassard
Environ 163 x 580 mm
Unité : Environ 207 g, piles exclues
Poids:
Brassard: Environ 128 g, Grand brassard : Environ 134 g
Température
10°C – 40°C
Conditions d’utilisation :90% RH
Humidité
Température–20°C – 55°C
Conditions de stockage :90% RH
HumiditéProtection contre les décharges électriques :Unité d’alimentation interne
Niveau de protection
Pièce de type BF appliquée sur le patient
Mode de fonctionnement :Fonctionnement continu
Mémoire :1 × 99 lectures, moyenne des 3 dernières lectures
Durée de vie :10 000 mesures ou 3 ansDurée de vie du brassard
Environ 1 000 utilisations
Degré d’étanchéité :IPX0
Ensemble composé d’un brassard, 4 piles AA (LR6)Accessoires :pour l’appareil et mode d’emploi
Brassard 22 cm – 30 cm, Grand brassard 22 cm – 42 cm
Accessoires en option :* Le brassard est la pièce appliquée sur le patient de ce dispositif.
* Plage de pression barométrique : 800 hPa à 1050 hPa* Cet appareil satisfait aux normes ci-dessous: IEC 60601-1/EN 60601-1 (Appareilsélectromédicaux – Partie 1), IEC 60601-1-2/EN 60601-1-2 (Appareils électromédicaux –Partie 1-2), EN 1060-3 (Tensiomètres non invasifs – Partie 3), IEC 80601-2-30 (Appareilsélectromédicaux – Partie 2-30)

EnglishSPÉCIFICATIONS
Numéro du modèle :Système de mesure :Affichage
Emplacement de la mesure
Brassard :Plage de circonférence du brassard :Pression du brassard :- Fra 16 -8/3/2024 9:21:31 AM

Русский
СодержаниеEnglishEspañol
Français

1 - ОБЩИЕ СВЕДЕНИЯ
3 - МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ ИОБСЛУЖИВАНИИ
6 - НАЗВАНИЯ И РАСПОЛОЖЕНИЕ ДЕТАЛЕЙ
7 - УСТАНОВКА БАТАРЕЕК
7 - ИЗВЛЕЧЕНИЕ БАТАРЕЕК
8 - НАСТРОЙКА ВРЕМЕНИ
9 - ИЗМЕРЕНИЕ АРТЕРИАЛЬНОГО ДАВЛЕНИЯ
12 - СОВЕТЫ ПО ПРАВИЛЬНОМУ ВЫПОЛНЕНИЮ ИЗМЕРЕНИЙ
12 - ВЫЗОВ ДАННЫХ, СОХРАНЕННЫХ В ПАМЯТИ
13 - ОБ “АРТЕРИАЛЬНОМ ДАВЛЕНИИ”
14 - ВОПРОСЫ И ОТВЕТЫ ОТНОСИТЕЛЬНО АРТЕРИАЛЬНОГО ДАВЛЕНИЯ
15 - ПРЕЖДЕ ЧЕМ ОБРАЩАТЬСЯ ЗА РЕМОНТОМ ИЛИ ПРОВЕРКОЙ
16 - ТЕХНИЧЕСКИЕ ХАРАКТЕРИСТИКИ

Pyсский

■ Убедитесь в наличии всех компонентов, перечисленных ниже.
• Прибор для измерения артериального давления • Инструкция по эксплуатации
• 4 батарейки типа AA (для прибора) • Манжета (модель: SCN-008) для прибора
CHU304 • Большая манжета (модель: SCW-009) для прибора CHU305

ОБЩИЕ СВЕДЕНИЯ
Общие сведения об артериальном давлении и его измерении
1.
2.

हिन्दी

3.
4.
5.
6.

Чтобы быстро выпустить воздух из манжеты, нажмите кнопку “START/STOP”.
Перед использованием манжеты для измерения давления, убедитесь, что выправильно подобрали размер манжеты:Манжета (модель: SCN-008) подходит для руки с обхватом: 22 см – 30 см.
Большая манжета (модель: SCW-009) подходит для руки с обхватом: 22 см – 42 см.
Надевая манжету, расположите ее таким образом, чтобы воздушный шланг
находился по центру руки, а край манжеты – на расстоянии 1 см – 2 см вышелоктя.
Самостоятельное измерение давления не заменяет лечения! Изменение
дозировки лекарств следует производить только по указанию врача.
Перед измерением артериального давления следует опорожнить мочевой
пузырь и спокойно посидеть от 10 до 15 минут.
НЕ преувеличивайте значение результатов единичного измерения.
Записывайте изменения артериального давления. Полную картину может датьтолько большое количество собранных данных.

Как правильно измерять артериальное давление
1.
2.
3.

Перед измерением давления сделайте пять или шесть глубоких вдохов ивыдохов и расслабьтесь. Если вы напряжены во время измерения, результат
будет неправильным.
Артериальное давление может подняться, если вы чем-то обеспокоены,раздражены, не выспались, страдаете запором, а также после выполнения
физических упражнений или приема пищи.
Оберните манжету вокруг руки и затяните так, чтобы между манжетой и рукой
помещался один палец. (См. стр. 9, 10)00_CHU304_305_2401.indb 1- Рус 1 -8/3/2024 9:21:32 AM

Гипертония, степень 3 (тяжелая)Примечания:График не является точным,но может использоватьсяв качестве ориентира дляпонимания результатов
неинвазивного измерения
артериального давления.
Прибор предназначен длявзрослых пользователей.

Гипертония, степень 2 (умеренная)Гипертония, степень 1 (слабая)Повышенное
Нормальное
Оптимальное

Español
FrançaisСИСТОЛИЧЕСКОЕ (mmHg)

PyсскийНЕ следует измерять артериальное давление после приема ванны илиупотребления спиртных напитков.
5. Измерять давление рекомендуется в помещении при температуре около
20°C. НЕ измеряйте давление при слишком низкой (ниже 10°C) или слишком
высокой (выше 40°C) температуре в помещении.
6. НЕ измеряйте артериальное давление после употребления кофе, чая иликурения.
7. Измеряйте артериальное давление в расслабленном, спокойном состоянии.
Центр манжеты должен находиться на уровне сердца. НЕ двигайте рукой и НЕразговаривайте.
8. НЕ следует измерять артериальное давление спустя несколько минут илиболее после надевания манжеты. В это время давление в руке повысится ирезультаты измерения будут неправильными.
9. В случае отображения индикатора нерегулярного сердцебиения (IHB)
постарайтесь расслабиться и проведите повторное измерение артериального
давления. В случае частого появления индикатора нерегулярного
сердцебиения (IHB), рекомендуется проконсультироваться с врачом по поводу
состояния здоровья. (См. стр. 11)
10. На следующем рисунке показана классификация артериального давления согласно
стандартам ВОЗ.

English4.

* Данный прибор оснащен индикатором, который визуально показывает
классификацию результатов артериального давления (оптимальное /нормальное / повышенное / гипертония, степень 1 / гипертония, степень 2 /гипертония, степень 3) после каждого измерения.
11. Предполагаемое использование
•Д анное устройство предназначено для неинвазивного измерения
артериального давления осциллометрическим способом. Устройство
может автоматически измерять систолическое давление крови (SYS),диастолическое давление крови (DIA) и частоту пульса.
•У стройство предназначено для проверки состояния здоровья в домашних
условиях и не является диагностическим прибором.
•У стройство не предназначено для использования на новорожденных илидетях и не может использоваться для автоматического измерения циклов.

हिन्दी

ДИАСТОЛИЧЕСКОЕ (mmHg)

Определение и классификация уровней артериального давления

Рекомендуется регулярно измерять артериальное давление водно и то же время.

* Артериальное давление постоянно изменяется. Это означает, что данные,собранные за продолжительный период времени, имеют большую значимость,чем единичное измерение. Поэтому следует измерять артериальное давление
ежедневно. В идеале, рекомендуется измерять артериальное давление в одно ито же время каждый день.

00_CHU304_305_2401.indb 2- Рус 2 -8/3/2024 9:21:32 AM

МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ И ОБСЛУЖИВАНИИ
Меры предосторожности при использовании
1.

EnglishEspañol
FrançaisPyсский
हिन्दी

Если вы страдаете заболеваниями сердца, гипертонией или другими
сосудистыми заболеваниями, сначала проконсультируйтесь со своим врачом.
2. Чрезмерное накачивание манжеты может привести к образованию синяка на руке.
Если вы почувствуете ненормальное давление в манжете или какиелибо другие нарушения в работе во время использования манжеты,немедленно уменьшите давление путем нажатия кнопки “START/STOP”,
снимите манжету или отсоедините воздушный шланг от прибора, а затем
обратитесь за консультацией по месту покупки прибора.
3. Если вы полагаете, что полученные значения измерений являются
ненормальными или при измерении вы чувствуете себя нехорошо,обратитесь к своему врачу.
4. Измерение артериального давления у людей со слабым пульсом илиаритмией может оказаться невозможным.
5. Многократное измерение давления и длительное чрезмерное накачивание унекоторых людей может привести к таким проблемам, как покраснение и отечность.
6. При повторных часто выполняемых измерениях артериального давления
могут быть получены неправильные результаты. Перед повторным
измерением следует подождать не менее 1 минуты.
7. Проконсультируйтесь со своим врачом перед использованием прибора, есливы страдаете серьезными нарушениями кровообращения в области рук.
Невыполнение этого требования может представлять опасность для здоровья.
8. Измерение артериального давления при недостаточном кровотоке в области выполнения
измерения или при частом либо нерегулярном сердцебиении может оказаться
невозможным. Проконсультируйтесь со своим врачом, можно ли использовать прибор.
9. НЕ надевайте манжету на поврежденную руку.
10. НЕ надевайте манжету на запястье под капельницей (во время
выполнения внутривенной инфузии) или при переливании крови. Этоможет привести к травме или другим тяжелым последствиям.
11. Не используйте одну и ту же манжету вместе с другими людьми, чтобы
избежать перекрестной инфекции.
12. При наличии сомнений относительно применения манжеты на руке с тойстороны, где была произведена мастэктомия, проконсультируйтесь с врачом.
13. НЕ используйте прибор в помещениях с воспламеняющимися газами,например такими, что используются для анестезии. Это может привестик воспламенению газов и стать причиной взрыва.
14. НЕ используйте прибор в помещениях с повышенным содержанием
кислорода, таких как барокамера или кислородная палатка. Это может
привести к воспламенению кислорода и стать причиной пожара.
15. НЕ пользуйтесь мобильным телефоном вблизи прибора, так как это может
стать причиной его неисправности.
16. Лицам, использующим кардиостимуляторы, следует предварительно
проконсультироваться с врачом.
17. Этот прибор должен использоваться только для измерения
артериального давления. НЕ используйте его для других целей.
18. Обязательно используйте только разрешенные детали ипринадлежности. Детали и принадлежности, не одобренные дляиспользования с данным прибором, могут привести к его повреждению.
19. НЕ используйте данный прибор для измерения давления у детей.
20. Измерение артериального давления может быть невозможным длялюдей с общими аритмиями, такими как предсердная или желудочковая
экстрасистолия или мерцательная аритмия.

00_CHU304_305_2401.indb 3- Рус 3 -8/3/2024 9:21:32 AM

5.
6.

7.
8.
9.

Внимание

Модификации данного прибора недопустимы.

Объяснение символов: Маркировка CE свидетельствует о соответствии устройства
основным требованиям директив ЕС.
: Используется деталь типа BF (Манжета является изделием типа BF.)
: Хранить в сухом месте
: Внимание
:П еред использованием обратитесь к инструкции по эксплуатации
(Цвет фона знака: синий. Графический символ знака: белый.): Устройство соответствует директиве WEEE
: Производитель
: Символ для “УПОЛНОМОЧЕННОГО ПРЕДСТАВИТЕЛЯ В ЕС”Внимание

Español

4.

Français3.

Pyсский

2.

НЕ храните прибор для измерения артериального давления в местах,подверженных воздействию прямого солнечного света, высоких
температур (выше 55°C), низких температур (ниже -20°C), высокой
относительной влажности (выше 90%) или чрезмерного количества пыли.
Храните прибор для измерения артериального давления в месте,недоступном для детей, домашних животных или насекомых.
НЕ допускайте падения прибора для измерения артериального давленияили манжеты и не подвергайте их другим ударам или вибрации.
Извлекайте батарейки, если прибор не используется в течение
длительного периода времени.
НЕ разбирайте прибор или манжету. Обратитесь в сервисный центр,указанный производителем.
НЕ сгибайте манжету и воздушный шланг слишком сильно.
В случае сильного загрязнения прибора или манжеты очистите их тканью,смоченной в спиртовом дезинфицирующем растворе или нейтральном
моющем средстве. Затем протрите их сухой тканью. При использовании вбольнице нужно проводить дезинфекцию манжеты дважды в неделю.
НИКОГДА не используйте для очистки прибора для измерения
артериального давления спирт, разбавитель или бензин, поскольку этоможет привести к повреждению прибора.
Для очистки манжеты протирайте ее влажной тканью. Избегайте
чрезмерного трения манжеты, поскольку это может стать причиной утечки
воздуха. Следите за тем, чтобы вода не попала в воздушный шланг.
Рекомендуется очищать манжету после каждого 200-кратного использования.

हिन्दी

1.

EnglishМеры предосторожности при уходе за прибором

Прибор прошел испытания в соответствии состандартом EN60601-1-2 для EMD. Это никоим
образом не гарантирует неподверженность устройства
электромагнитным помехам. Избегайте использования
устройства в среде электромагнитного излучения.

Характер и частота обслуживания:Данный прибор предназначен для использования в течение длительного
периода времени;однако, обычно рекомендуется проводить проверку каждые два года дляобеспечения надлежащего функционирования и эффективности.

00_CHU304_305_2401.indb 4- Рус 4 -8/3/2024 9:21:32 AM

Защита окружающей среды:Оказывайте содействие в защите окружающей среды, соблюдая
государственные законы и/или местные правила при утилизации батареек иизделия по окончании их срока службы.

ЗНАК WEEE
EnglishВ случае утилизации данного прибора не выбрасывайте его вместес обычными бытовыми отходами. Существует система раздельного
сбора использованных электронных изделий в соответствии сзаконодательством по реализации Директивы WEEE (Директива
2012/19/EU), которая имеет силу только в Европейском Союзе.

EspañolЕсли вам необходимо утилизировать использованные батареи, обратитесь в местные
органы власти или к продавцу, и спросите о правильном способе утилизации.

FrançaisИнформация относительно утилизации в других странах запределами Европейского Союза.

Европейский стандарт:Примечание относительно символа батареи.

Данный символ может использоваться в сочетании с символом химического
вещества. В этом случае к нему применяется требование, предусмотренное
Директивой относительно данного химического вещества.
Данный прибор для измерения артериального давления соответствует стандарту EN60601-1, EN1060-1, EN1060-3, EN1060-4.

Pyсский

Калибровка:Прибор для измерения артериального давления рекомендуется
проверять каждые два года, чтобы быть уверенным в правильности егоработы и точности показаний. Обратитесь к дистрибьютору.

हिन्दी

Наименование :Адрес :

CITIZEN SYSTEMS JAPAN CO., LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, JapanНаименование :Адрес :ANDON HEALTH CO., LTD.
No.3 Jinping Street, YaAn Road, Nankai District, Tianjin
300190, China

ЗаводПредставительство в Европе:Наименование :Адрес :EMERGO EUROPE
Westervoortsedijk 60, 6827 AT Arnhem,Нидерланды• Для получения информации о технических
данных обратитесь к европейскому представителю
компании.

Сделано в Китае

00_CHU304_305_2401.indb 5- Рус 5 -8/3/2024 9:21:32 AM

НАЗВАНИЯ И РАСПОЛОЖЕНИЕ ДЕТАЛЕЙ
Названия частей
Воздушный шланг

Дисплей

КнопкаMEMORYМеталлическая
петляРазъем
воздушного
шланга

FrançaisКнопка START/STOPОжидание

Pyсский

Гнездо для подсоединения
воздушного шланга
* Для получения подробной информации о батарейном отсеке см. стр. 7.

<Дисплей>Значок батареек

Индикация готовностик нагнетанию

Индикация
систолического
артериального
давления
Индикатор
классификации
гипертонии

Español

English<Манжета>हिन्दी

<Основной блок>Значок
памяти
Индикация
диастолического
артериального
давления/пульса

Индикатор нерегулярного Индикация даты/времени
сердцебиения (IHB)** Для получения подробной информации об индикаторе нерегулярного
сердцебиения (IHB) см. стр. 11.

00_CHU304_305_2401.indb 6- Рус 6 -8/3/2024 9:21:33 AMУСТАНОВКА БАТАРЕЕК
(Перед использованием прибора для измерения артериального давления
необходимо установить батарейки, поставляемые с прибором, в батарейный отсек).

English1 Откройте крышку батарейного отсека.
Чтобы открыть отсек, нажмите на верхнюю часть крышки и потяните еена себя.
Батарейный отсек

EspañolКрышка батарейного
отсека

Français* Если прибор для измерения артериального давления неиспользуется в течение длительного периода времени, следует
извлечь батарейки из отсека.

2 Установите батарейки в отсек.

Pyсский

Установите батарейки в отсек,учитывая символы полярности

и.

4 батарейки типа “AA”

3 Закройте крышку батарейного отсека.

हिन्दी

ИЗВЛЕЧЕНИЕ БАТАРЕЕК
• При отображении символаили в случае отсутствия изображенийна дисплее, замените старые батарейки четырьмя новыми. Все четыре
батарейки следует заменять одновременно.
• Выполните настройку часов после замены батареек.
• Данные, сохраненные в памяти, не удаляются при замене батареек.
• Не используйте перезаряжаемые батарейки.
• Не используйте одновременно сухие щелочные (LR6) и марганцевые
батарейки (R6P).
• Прилагаемые батарейки предназначены только для демонстрационных
целей. Срок службы батареек может быть короче указанного.
• При утилизации использованных батареек соблюдайте государственные
нормы или стандарты общественных институтов по охране окружающей
среды, действующие в вашей стране/вашем регионе.

00_CHU304_305_2401.indb 7- Рус 7 -8/3/2024 9:21:33 AMНАСТРОЙКА ВРЕМЕНИMEMORY” в течение
START/STOP” и кнопку “примерно двух секунд, пока не услышите
звуковой сигнал и на экране не начнет мигать
индикатор “месяц”.

3 Нажмите кнопку “

MEMORY” для установкиMEMORY” длязначения “месяц”. Нажмите кнопку “увеличения значения на единицу. Нажмите кнопку “ START/STOP” для подтверждения установки. Значение “месяц” будет
установлено, и начнет мигать индикатор “день”.
* Отображение значений можно ускорить, нажав и
MEMORY.
удерживая нажатой кнопку

Español

2 Нажмите и удерживайте нажатой кнопку “Françaisпо очереди будут отображаться время и дата.

Pyсский

1 При установке батареек в первый раз на прибореEnglishУстановка часов обеспечивает точную запись измерений.
Прибор может записывать дату и время измерения, а также результаты
измерений. Обязательно установите часы после установки батареек.

MEMORY” для установкиMEMORY” длязначения “день”. Нажмите кнопку “увеличения значения на единицу. Нажмите кнопку “
START/STOP” для подтверждения установки. Значение
“день” будет установлено, и начнет мигать индикатор “час”.

हिन्दी

4 Нажмите кнопку “
5 Нажмите кнопку “

MEMORY” для установкиMEMORY” длязначения “час”. Нажмите кнопку “увеличения значения на единицу. Нажмите кнопку
“ START/STOP” для подтверждения установки.
Значение “час” будет установлено, и начнет мигать
индикатор “минуты”.

6 Нажмите кнопку “

MEMORY” для установкиMEMORY”значения “минуты”. Нажмите кнопку “для увеличения значения на единицу. Нажмите
кнопку “ START/STOP” для подтверждения
установки.
* Нажмите кнопку “ START/STOP” одновременнос сигналом точного времени по радио и т.п. длязавершения настойки часов.

00_CHU304_305_2401.indb 8- Рус 8 -8/3/2024 9:21:34 AMИЗМЕРЕНИЕ АРТЕРИАЛЬНОГО ДАВЛЕНИЯ➊ Вставьте разъем воздушного
шланга в основной блок.

English• Не сгибайте воздушный шланг вовремя измерений, поскольку этоРазъем воздушного Гнездо дляможет привести к ошибке нагнетания
шланга подсоединенияили причинить травму в связи своздушного
постоянным давлением манжеты.
шланга

Español➋ НАДЕВАНИЕ МАНЖЕТЫ

FrançaisПеред использованием манжеты для измерения давления, убедитесь,что вы правильно подобрали размер манжеты:Манжета (модель: SCN-008) подходит для руки с обхватом: 22 см – 30 см.
Большая манжета (модель: SCW-009) подходит для руки с обхватом:22 см – 42 см.

1Pyсский

2Разверните манжету и пропустите ее конец
сквозь металлическую петлю таким образом,чтобы сторона с застежкой-липучкой
оказалась снаружи.
Наденьте манжету на руку таким образом,чтобы воздушный шланг был расположен состороны ладони.
Металлическая
петляहिन्दी

3: Положение артерии.

Расположите манжету таким образом, чтобы воздушный шланг
находился по центру руки, а край манжеты – на расстоянии 1 см – 2 смвыше локтя.

00_CHU304_305_2401.indb 9* Манжету следует надевать на голую
руку или поверх легкой одежды.
* Если на вас надета одежда изплотной ткани, снимите ее.
1 см – 2 см- Рус 9 -8/3/2024 9:21:34 AM

EnglishEspañolПоложите руку на стол или инуюповерхность таким образом, чтобы
центр манжеты находился на уровне
вашего сердца.

6

Français* Располагая манжету на руке,убедитесь в том, что она правильно
затянута. Натяжение манжеты
считается правильным, если между
манжетой и рукой можно свободно
пропустить палец.

Расслабьте свою руку и слегка откройте ладонь вверх.

➌ Разместите манжету на уровне сердца.
➍ Нажмите кнопку “

START/STOP”.

• Раздастся звуковой сигнал и на начальном экране загорятся всецифры, свидетельствующие о должном функционирование прибора.
• Манжета накачивается автоматически.
* Если устройство фиксирует недостаточное значение накачивания,оно автоматически повторяет накачивание воздуха.

Pyсский

5Потяните конец манжеты наружу
таким образом, чтобы манжета
плотно облегала руку, а затем
зафиксируйте его с помощью
застежки-липучки.

हिन्दी

4*	 отображается при обнаружении пульса.
*	Если необходимо остановить измерение, нажмите кнопку “ START/STOP”. Воздух из манжеты выпускается, и измерение прекращается.
* Если предыдущий пользователь задал высокое значение давления,оно также будет использовано и для текущего измерения.
* Если значение накачивания превышает 260 мм рт.ст.,накачивание причиняет дискомфорт или если необходимо
прекратить процесс измерения, еще раз нажмите кнопку “
START/STOP”. Манжета выпустит воздух, и питание отключится.

00_CHU304_305_2401.indb 10- Рус 10 -8/3/2024 9:21:34 AM➎ Отображаются результаты измерений.
•EnglishИндикатор классификации
гипертонии

Español➏ Завершение измерения.

FrançaisПо завершении измерения
манжета выпускает воздух, а надисплее отображаются результаты
измерения (значения систолического/диастолического артериального
давления, пульса).
* Значения систолического/диастолического артериального
давления и пульса отображаются
попеременно.
* При условии отсутствия ошибки врезультатах измерения, такие значения
автоматически сохраняются в памяти.

Индикатор нерегулярного сердцебиения

• Чтобы отключить прибор, нажмите кнопку “ START/STOP”.
• После выполнения измерения артериального давления нажатие кнопки
“ START/STOP” выключит прибор. Однако если в течение 1 минуты небудет нажата ни одна кнопка, прибор будет автоматически выключен.

PyсскийВ случае обнаружения нерегулярного сердцебиения во время
измерения, после завершения измерения отображается индикатор
нерегулярного сердцебиения (IHB).
Индикатор нерегулярного сердцебиения (IHB) отобразиться при 2 условиях:1) Коэффициент вариации (CV) пульсового периода составляет > 25%2) Р азность соседних пульсовых периодов составляет 0,14 с, а ихколичество – более 53 процентов от общего пульса.
• В случае значительных колебаний частоты сердечных сокращений во время измерения,его результаты могут быть неверными. В случае отображения индикатора нерегулярного
сердцебиения (IHB), проведите повторное измерение артериального давления.

हिन्दी

ОсторожноВ случае частого появления индикатора нерегулярного
сердцебиения (IHB), рекомендуется проконсультироваться сврачом по поводу состояния здоровья.

Индикатор классификации гипертонии

Значение измеренного артериального давления отображается согласно стандартам ВОЗ.
(См. стр. 2)

00_CHU304_305_2401.indb 11СИСТОЛИЧЕСКОЕ (mmHg)
180
160
140
130
120

Гипертония, степень 3 (тяжелая)Гипертония, степень 2 (умеренная)Гипертония, степень 1 (слабая)Повышенное
Нормальное
Оптимальное
80

85

90 100- Рус 11 -110 ДИАСТОЛИЧЕСКОЕ
(mmHg)

8/3/2024 9:21:35 AMВ качестве технических аварийных сигналов на ЖК-дисплее будут немедленно отображаться
значения «HI» или «Lo», если измеряемое артериальное давление (систолическоеили диастолическое) выходит за рамки рабочего диапазона, указанного в разделе
«ТЕХНИЧЕСКИЕ ХАРАКТЕРИСТИКИ». В таком случае вам следует проконсультироваться сврачом или проверить, использовался ли прибор без нарушения инструкций.
Техническая аварийная ситуация (выход за рамки рабочего диапазона) предварительно
устанавливается на заводе и не может быть настроена или отключена. Данной аварийной
ситуации присвоен низкий приоритет согласно IEC 60601-1-8. Технический сигнал опасности
является нефиксированным и не требуется сброса. Сигнал, отображающийся на ЖК-дисплее,исчезнет примерно через 8 секунд.

EnglishОписание технических аварийных сигналов

ВЫЗОВ ДАННЫХ, СОХРАНЕННЫХ В ПАМЯТИВ памяти можно сохранить 99 результатов
измерения.
Среднее значение вычисляется
автоматически, что помогает ежедневно
контролировать состояние здоровья.

1

FrançaisPyсский
हिन्दी

■ Для измерения артериального давления можно использовать правую руку.
Значение артериального давления при измерении на левой руке может
отличаться на 10 мм рт.ст. от измерений на правой руке (или наоборот).
Измеряйте артериальное давление на одной и той же руке каждый день.

EspañolСОВЕТЫ ПО ПРАВИЛЬНОМУ ВЫПОЛНЕНИЮ ИЗМЕРЕНИЙ
Артериальное давление отличается в зависимости от положения, времени сутоки множества других факторов. В идеале, рекомендуется измерять артериальное
давление в одном и том же положении в одно и то же время каждый день.
■ Измерение в положении сидя*	Сядьте удобно, не скрещивая ноги. Стопы должны стоять на полу.
➀ Положите руку на стол или на аналогичную
поверхность, вытянув предплечье.
➁ Манжета должна располагаться на уровне сердца.
➂ Положите руку на стол ладонью вверх ирасслабьте пальцы.
➃ Не двигайтесь и не разговаривайте во время
измерения.
■ Измерение в положении лежа* Расслабьтесь, чтобы несоздавать давления на манжету.

Кнопка
памяти

Нажмите один раз кнопку памяти
.
*	Пример индикации при нажатии кнопки памяти.
На ЖК-дисплее отображается суммарное значение измерений, аотображает среднее значение в нижней части экрана. Затем будет
отображено среднее значение на основе последних трех измерений.

00_CHU304_305_2401.indb 12- Рус 12 -8/3/2024 9:21:35 AM

2При повторном нажатии кнопки памяти
будут отображены
данные предыдущего измерения.
При каждом нажатии кнопки будут отображаться данные измерений впорядке от наиболее новых к наиболее старым.
.
* Пример индикации при нажатии кнопки памятиEnglishEspañolПеред результатами
измерений будет
отображаться номер данных.

Français• Значения, сохраненные в памяти, нумеруются в порядке проведения
измерений. Например, в случае сохранения в памяти 99 наборов данных,номер данных 1 будет представлять самые последние данные.
Номер данных: 1, 2, 3, ...
... 98, 99ÇÇСамые последние данные Более ранние данныеКАК УДАЛИТЬ ДАННЫЕ

Pyсский

1Во время отображения результатов (за исключением среднего
показателя последних трех измерений) нажмите и удерживайте
.
более 3 секунд кнопку MemoryВсе данные, сохраненные в памяти, будут удалены.

हिन्दी

Нажмите иудерживайте
нажатой втечение 3 секунд
Удалено

Будет отображено сообщение,информирующее о том, что вседанные были удалены из памяти.

ОБ “АРТЕРИАЛЬНОМ ДАВЛЕНИИ”Что такое артериальное давление? (Систолическое)(Диастолическое)Сердце представляет собой
Артерия
Артерия
насос, с помощью которого
Левое
Левое
предсердие
предсердие
кровь циркулирует по телу.
Кровь при постоянном давлении
Стенка артерии
прокачивается через сердце в
Левый
Левый
возвращается вжелудочек Стенка артерии
желудочек
артерии. Это давление называется
нормальное состояние.
расширяется.
артериальным и отражает, в общем,Расширение
Сжатие
давление крови. Артериальное
давление характеризуется
несколькими параметрами, такими как систолическое давление, возникающее, когда
сердце качает кровь, и диастолическое давление, возникающее, когда кровь возвращаетсяв сердце.

00_CHU304_305_2401.indb 13- Рус 13 -8/3/2024 9:21:36 AM

140
120
100
80
60
40

Диастолическое
артериальное
давлениеВо сне
12 13 14 15 16 17 18 19 20 21 22 23 24 1

2Время суток

3

4

5

6

7

8

(Артериальное давление,измеренное с пятиминутным
интервалом в течение всего
дня)●B
 evan AT, Honour AJ, StottFH. Clin Sci 1969;36:329–44

9 10 11ВОПРОСЫ И ОТВЕТЫ ОТНОСИТЕЛЬНО АРТЕРИАЛЬНОГО ДАВЛЕНИЯ

Q Почему артериальное давление, измеренное дома, отличается отартериального давления, измеренного врачом или в больнице?

A Артериальное давление зависит от физического состояния, температуры

окружающей среды или психического состояния. Когда врач или медсестра
измеряет артериальное давление, показания давления, скорее всего, будут
выше обычного на от 10 до 20 мм рт.ст. вследствие взволнованности и/илистресса. Знание факторов, влияющих на артериальное давление, поможетвам лучше следить за состоянием вашего здоровья.

Q Почему измеренное артериальное давление отличается при каждом

EspañolСистолическое
артериальное
давление

160

FrançaisЗначение артериального
давления (мм рт.ст.)[Пример изменений артериального давления в течение дня]Pyсский

Артериальное давление отличается в зависимости от возраста, пола и рядадругих факторов. На него оказывают влияние дневные биоритмы, положение
тела, физическая деятельность, психическая активность, уровень стресса идаже температура воздуха. В течение дня артериальное давление здорового
человека обычно меняется.

EnglishАртериальное давление постоянно изменяется.

Артериальное давление отличается при каждом сердечном сокращении. Мыдумаем, что наше давление является постоянным, однако оно изменяется,даже если проводить измерения повторно. Артериальное давление
чувствительно ко времени дня, времени года и температуре воздуха. Кроме
того, на артериальное давление оказывают влияние психические стрессы,эмоциональные подъемы и спады. Оно увеличивается, когда вы напрягаетесь,и снижается, когда расслабляетесь.

हिन्दी

измерении?

A Наше артериальное давление автоматически регулируется нервной системой.

Q Какие преимущества дает измерение артериального давления дома?
A Измерение артериального давления дома дает более достоверные

результаты, так как вы можете выполнять измерение, когда находитесь вспокойном и расслабленном состоянии. Врачи придают большое значение
показаниям замеров артериального давления в домашних условиях. Следует
измерять артериальное давление в одно и то же время каждый день и вести
записи изменений в показаниях замеров артериального давления.

Всегда обращайтесь к врачу за консультацией по поводу анализа
результатов измерения артериального давления и назначением
требуемого лечения.

00_CHU304_305_2401.indb 14- Рус 14 -8/3/2024 9:21:36 AMПРЕЖДЕ ЧЕМ ОБРАЩАТЬСЯ ЗА РЕМОНТОМ ИЛИ ПРОВЕРКОЙ
Перед тем как отправить прибор для измерения артериального давления напроверку или для ремонта, проверьте следующее.

EnglishУстранение неисправностей (1)Неисправность

EspañolНа ЖК-дисплее
отображаются
нехарактерные
результаты

Возможная причина

Неправильное положение манжеты илиона ненадлежащим образом затянута
Неправильное положение тела вовремя проведения измерений
Разговор, движения рукой иликорпусом, чувство тревоги,возбуждения или нервное состояниево время проведения измерений

FrançaisНерегулярное сердцебиение
(аритмия)Решение

Надлежащим образом расположите
манжету и проведите повторное измерение.
Еще раз ознакомьтесь с разделом
“ПОЛОЖЕНИЕ ТЕЛА ПРИПРОВЕДЕНИИ ИЗМЕРЕНИЙ”данного руководства и проведите
повторное измерение.
Повторное измерение следует
проводить молча, в спокойном
состоянии, без лишних движений.
Использование электронного
сфигмоманометра лицами,страдающими тяжелыми формами
аритмии, запрещено.

Устранение неисправностей (2)Неисправность

Возможная причина

Решение

Pyсский
हिन्दीНа ЖК-дисплее
отображается символ
Низкий уровень заряда батареек
Замена батареек.
низкого заряда батареекНа ЖК-дисплее
Нестабильное
давление
передотображается сообщение
проведением измеренийоб ошибке “Er 0”На ЖК-дисплее
Необнаруживается
систолическое
Постарайтесь не двигаться воотображается сообщение
артериальное давление
время повторного измерения.
об ошибке “Er 1”На ЖК-дисплее
диастолическое
отображается сообщение Не обнаруживается
артериальное давлениеоб ошибке “Er 2”На ЖК-дисплее
Работа пневматической системы
отображается сообщение
заблокирована или манжета
Надлежащим образом расположитеоб ошибке “Er 3”слишком затянута для нагнетания
манжету и проведите повторноеНа ЖК-дисплее
Утечка в пневматической системе
измерение.
отображается сообщениеили слишком слабая затяжкаоб ошибке “Er 4”манжеты для нагнетанияНа ЖК-дисплее
отображается сообщение Давление манжеты выше 300 мм
рт.ст.
об ошибке “Er 5”На ЖК-дисплее
Давление манжеты в течение 3Выполните повторное
отображается сообщение
минут остается выше 15 мм рт.ст.
измерение через пять минут.
об ошибке “Er 6”Если на дисплее по-прежнемуНа ЖК-дисплее
Ошибка доступа к электрически
отображаются нехарактерные
отображается сообщение
программируемому ПЗУпоказания, свяжитесь с местнымоб ошибке “Er 7”На ЖК-дисплее
дистрибьютором илиотображается сообщение Ошибка проверки параметров прибор
заводом-изготовителем.
об ошибке “Er 8”На ЖК-дисплее
Неверный параметр датчика
отображается сообщение
давленияоб ошибке “Er A”
Устройство не отвечает
Извлеките батарейки на пять минут,Неправильная
эксплуатация илипри нажатии кнопок или сильные электромагнитныеих обратно напомехи а затем установите
установке батареек.
место.

* В случае утилизации данного изделия не выбрасывайте его вместе с обычными
бытовыми отходами. В соответствии с законодательством для использованных
электронных изделий предусмотрены системы раздельного сбора.

00_CHU304_305_2401.indb 15- Рус 15 -8/3/2024 9:21:36 AMКнопка:Нагнетание:Спуск:
Номинальное напряжение:Выпуск:Источник питания:Срок службы батареек:Функция автоматического выключения:Размеры основного блока:Манжета
Размеры манжеты:Большая манжета
Вес:Температура
Относительная
влажность
Температура
Условия хранения:Относительная
влажность
Защита от поражения электрическим током:Степень охраны
Режим работы:Память:Срок службы:Ресурс манжеты:Защита от проникновения воды:Условия эксплуатации:Принадлежности:Дополнительные принадлежности:EspañolЗначки

FrançaisИндикация на ЖКД:Pyсский

Погрешность:Давление
ПульсДавление
ПульсДавление
Пульс

90%-20°C – 55°C
90%Внутренний источник питания
Используются детали типа BF
Непрерывная работа
1 × 99 измерений, среднее значение последних 3 измерений
10 000 измерений или 3 годаПриблиз. 1000 измерений
IPX0В комплект поставки входит манжета, 4 батарейки типаАА (LR6) для прибора, инструкция по эксплуатации
Манжета – 22 см – 30 см, большая манжета – 22 см – 42 смहिन्दी

Диапазон измерения:CHU304, CHU305
Осциллометрический методТип цифрового дисплея
ПлечоМягкая манжета
22 см – 30 см (манжета), 22 см – 42 см (большая манжета)0 – 300 mmHg
Систолическое: 60 – 260 мм рт.ст.
Диастолическое: 40 – 199 мм рт.ст.
От 40 до 180 ударов/мин
±3 мм рт.ст.
±5% от показаний
3 цифры
3 цифры
: Значок измерения
: Индикация готовности к нагнетанию
: Значок батареи
: Индикатор нерегулярного сердцебиения (IHB): Значок памяти
: Индикатор классификации гипертонии
2 (START/STOP, MEMORY)Автоматическое нагнетание встроенным насосом
Автоматическая система быстрого спуска воздуха
6В постоянного тока(: постоянный ток)Электромагнитный клапан быстрого выпуска
4 батарейки 1.5 В, тип AA (LR6)Приблиз. 800 измерений (щелочные)Через приблиз. 1 мин (после активизации)138 (Ш) × 54 (В) × 95 (Г) ммПрибл. 148 × 510 ммПрибл. 163 × 580 ммАппарат: Приблиз. 207 г без батареек
Манжета: прибл. 128 г, большая манжета: прибл. 134 г
10°C – 40°C

EnglishТЕХНИЧЕСКИЕ ХАРАКТЕРИСТИКИ
Номер модели:Система измерения:Дисплей
Место измерения
Манжета:Обхват манжеты:Давление манжеты:* Рабочей частью данного устройства является манжета.
* Диапазон барометрического давления от 800 гПа до 1050 гПа* Это устройство соответствуют указанным ниже стандартам: IEC 60601-1/EN 60601-1(Медицинское электрическое оборудование – часть 1), IEC 60601-1-2/EN 60601-1-2(Медицинское электрическое оборудование – часть 1-2), EN 1060-3 (Неинвазивные
сфигмоманометры - часть 3), IEC 80601-2-30 (Медицинское электрооборудование –часть 2-30)

00_CHU304_305_2401.indb 16- Рус 16 -8/3/2024 9:21:36 AMअंग्रेज़ी
विषय-सचू ी

EnglishEspañol
Français

1 - सामान्य टिप्पणियाँ
3 - उपयोग और रखरखाव के लिए सावधानियाँ
6 - पुर्जों की पहचान
7 - बैट्री लगाना
7 - बैट्री निकालना
8 - समय समायोजन
9 - अपना ब्लड प्रेशर मापना
12 - सटीक माप प्राप्त करने की युक्तियाँ
12 - मेमोरी में संग्रहीत डेटा निकालना
13 - "ब्लड प्रेशर" के बारे में जानकारी
14 - ब्लड प्रेशर के बारे में प्रश्नोत्तर
15 - उपकरण की मरम्मत या जाँच के लिए अनुरोध करने से पूर्व के
16 - विनिर्दे श

Pyсский

■	यह सुनिश्चित कर लें कि आपके पास निम्नलिखित सभी घटक मौजूद हैं।• ब्लड प्रेशर मॉनिटर यूनिट • निर्दे श पुस्तिका
• 4 AA-साइज़ बैट्री (मॉनिटर के लिए)• CHU304 के लिए कफ़ (मॉडल: SCN-008)• CHU305 के लिए चौड़ा कफ़ (मॉडल: SCW-009)

सामान्य टिप्पणियाँ

ब्लड प्रेशर और ब्लड प्रेशर माप पर सामान्य टिप्पणियाँ

1.
2.

हिन्दी

3.
4.
5.
6.

कफ़ की वायु तेज़ी से निकालने के लिए “स्टार्ट /स्टॉप” स्विच दबाएँ।
अपना ब्लड प्रेशर कफ़ लगाने से पहले, यह सनु िश्चित करें कि आपने उचित आकारके कफ़ का चयन किया है :कफ़ (मॉडल: SCN-008) इस माप की बाँह में सही बैठता है : 22 सेमी – 30 सेमी।
चौड़ा कफ़ (मॉडल: SCW-009) इस माप की बाँह में सही बैठता है : 22 सेमी – 42 सेमी.
कफ़ लगाते समय, कफ़ की स्थिति इस तरह समायोजित करें कि वायु की नली आपकी बाँहके केंद्र में स्थित हो और कफ़ का किनारा आपकी कोहनी के 1 सेमी – 2 सेमी ऊपर हो।स्वयं का मापन उपचार नहीं है ! किसी भी स्थिति में आपको अपने चिकित्सक द्वारा
निर्धारित दवा की खुराक में बदलाव नहीं करना चाहिए।
अपना ब्लड प्रेशर मापने की तैयारी के लिए, आपको मूत्र-त्याग कर लेना चाहिए औरफिर मापने के पहले 10 से 15 मिनट तक स्थिर रहना चाहिए।
केवल एक बार के माप के परिणाम को ही अत्यधिक महत्व न दें । अपने ब्लड प्रेशरके निरं तर परिवर्तन का रिकॉर्ड रखें। कई बार की रीडिंग द्वारा ही समग्र अनुमान लगाया जा सकता है ।ब्लड प्रेशर का सही माप प्राप्त करने के लिए
1.
2.
3.
4.

अपना ब्लड प्रेशर मापने से पहले पाँच या छः बार गहरी सांस लें और तनाव रहित हो जाएँ।अगर ब्लड प्रेशर मापते समय आप तनाव में हैं, तो आपको सही रीडिंग नहीं मिलेगी।अगर आप बेचन ै या परे शान हैं, अनिद्रा या कब्ज़ से ग्रस्त हैं या फिर थोड़ी दे र पहलेव्यायाम किया है या भोजन लिया है , तो आपका ब्लड प्रेशर बढ़ा हुआ आएगा।
अपनी बाँह पर कफ़ को बांधें और उसे इस तरह कसें जिससे एक उं गली कफ़पषृ ्ठ 9, 10 दे खें)और आपकी बाँह के बीच में  प्रवेश कर सके। (स्नान करने या कुछ पीने के बाद अपना ब्लड प्रेशर न मापें ।

05_CHU304_Hin.indd 1- Hin 1 -6/3/2024 12:33:51 PM

सिस्टोलिक (mmHg)

श्रेणी 3 उच्च रक्तचाप (गंभीर)श्रेणी 2 उच्च रक्तचाप (मध्यम)श्रेणी 1 उच्च रक्तचाप (हल्का)उच्च-सामान्य
सामान्य
आदर्शडायस्टोलिक
(mmHg)

टिप्पणियाँ:यह ग्राफ सटीक नहीं है , लेकिन गैर-आक्रामक ब्लड प्रेशर के मापोंको जानने के लिए मार्गदर्शन केरूप में  प्रयोग किया जा सकताहै । इस डिवाइस का उपयोग
केवल वयस्क लोगों को ही करनाचाहिए।

ब्लड प्रेशर स्तर की परिभाषाएँ एवं वर्गीकरण
* यह यूनिट एक संकेतक से युक्त है , जो, प्रत्येक माप के बाद, परिणाम के ब्लड प्रेशर कावर्गीकरण (आदर्श / सामान्य / उच्च-सामान्य / श्रेणी 1 उच्च ब्लड प्रेशर / श्रेणी 2 उच्च ब्लड प्रेशर / श्रेणी 3 उच्च ब्लड प्रेशर) विजुअल रूप से इंगित करता है ।
11. अभीष्ट उपयोग
• यह डिवाइस दोलायमान विधि द्वारा शरीर में  प्रवेश किए बिना ब्लड प्रेशर कामॉनिटर है । यह डिवाइस स्वचालित रूप से सिस्टोलिक ब्लड प्रेशर (SYS), डायस्टोलिक ब्लड प्रेशर (DIA) और नाड़ी गति माप सकता है ।• इस डिवाइस का प्रयोजन चिकित्सक के मार्गदर्शन में घर पर निजी स्वास्थ्य की स्थिति की जाँच करना है , और इसका प्रयोजन नैदानिक डिवाइस का नहीं है ।
• इस डिवाइस का प्रयोजन नवजात शिशओु ं या शिशओु ं में उपयोग के लिए नहीं हैऔर इसका प्रयोजन स्वतः चक्रीय मापन के लिए नहीं है ।

EnglishEspañol
Françaisकॉफी अथवा चाय पीने या धूम्रपान के तुरंत बाद अपने ब्लड प्रेशर को मापने कीकोशिश न करें ।
7. अपना ब्लड प्रेशर तभी मापें  जब आप शांत और स्थिर हों। कफ़ के मध्य भाग को अपनेहृदय के समानांतर रखें और अपनी बाँह को बिलकुल भी न हिलाएँ और न ही बातचीत करें ।
8. अगर कफ़ आपकी बाँह पर कुछ मिनट या उससे अधिक दे र से बंधा हुआ हो, तोब्लड प्रेशर न मापें । इस दौरान आपकी बाँह में रक्त का दबाव बढ़ जाएगा औरआपको सही रीडिंग नहीं मिलेगी।
9. अगर अनियमित हृदय स्पंदन (IHB) का संकेतक प्रदर्शित होता है , तो आरामकरें और अपना ब्लड प्रेशर दोबारा मापें । अगर अनियमित हृदय स्पंदन (IHB) कासंकेतक बार-बार प्रदर्शित होता है , तो आपको अपने स्वास्थ्य की स्थिति के बारे मेंपषृ ्ठ 11 दे खें)चिकित्सक से परामर्श करने की सलाह दी जाती है । (10. नीचे दिए गए रे खा-चित्र में विश्व स्वास्थ्य संगठन (WHO) के मापदं डों के अनसु ार ब्लड
प्रेशर के वर्गीकरण दिखाए गए हैं।Pyсский

6.

अपना ब्लड प्रेशर वहाँ मापें  जहाँ कमरे का तापमान लगभग 20°C हो। अगरकमरे में अत्यधिक ठं ड (10°C से कम) या अत्यधिक गर्मी (40°C से अधिक)हो तो अपना ब्लड प्रेशर न मापें ।हिन्दी

5.

अपने ब्लड प्रेशर का माप प्रत्येक दिन एक ही समय पर लें।* आपका ब्लड प्रेशर हमेशा बदलता रहता है । इसका मतलब यह है कि जो आँकड़े लंबेसमय से लिए गए हैं, उनका महत्व सिर्फ एक बार के आँकड़े से कहीं अधिक है । इसकारण से, आपको अपने ब्लड प्रेशर का माप हर दिन लेना चाहिए। आदर्श रूप से,जहाँ तक संभव हो, आपको अपने ब्लड प्रेशर का माप प्रतिदिन एक ही समय पर लेनाचाहिए।

05_CHU304_Hin.indd 2- Hin 2 -6/3/2024 12:33:51 PM

उपयोग और रखरखाव के लिए सावधानियाँ

उपयोग के लिए सावधानियाँ
1.

EnglishEspañol
FrançaisPyсский
हिन्दीअगर आप हृदय रोग, उच्च ब्लड प्रेशर या किसी परिसंचरण संबंधी अन्य बीमारी सेपीड़ित हैं, तो मॉनिटर का उपयोग करने से पूर्व अपने चिकित्सक की सलाह लें।2. अधिक दबाव दे ने पर उस स्थान पर जख्म हो सकता है , जहाँ पर कफ़ बंधा होताहै । कफ़ का उपयोग करते समय अगर दबाव असामान्य लग रहा हो या आपकोकिसी
अन्य अनियमितता का अनुभव हो रहा हो, तो तुरंत “स्टार्ट /स्टॉप” बटन दबाकर दबावकम कर दें , पटटी को खोल लें या वायु नलिका को मॉनिटर से अलग कर दें औरफिर उस विक्रेता से संपर्क करें , जिससे आपने मॉनिटर खरीदा है ।
3. अगर आपको लगता है कि माप अस्वाभाविक है या फिर माप से आपको कोईअसुविधा हो रही है , तो उसका उपयोग रोक दें और अपने चिकित्सक की सलाह लें।4. उन व्यक्तियों के ब्लड प्रेशर का माप लेना शायद संभव न हो, जिनकी नाड़ी गति कम हो या जो अतालता से पीड़ित हों।5. ब्लड प्रेशर का बार-बार माप लेने और दे र तक अधिक दबाव पड़ने से कछ लोगों कोरक्त अवरोध या सजन जसै ी समस्या हो सकती है ।
ू
6. ब्लड प्रेशर के बार-बार मापन से एकदम सही परिणाम नहीं मिलेंगे। मापों के बीच में
कम-से-कम 1 मिनट का अंतराल रखें।
7. अगर आपको अपनी बाँह में रक्त संचार संबंधी गंभीर समस्या हो, तो मॉनिटर काउपयोग करने से पहले अपने चिकित्सक की सलाह लें। ऐसा नहीं करने से यह आपकेस्वास्थ्य के लिए हानिकारक हो सकता है ।
8. उस व्यक्ति का माप लेना संभव नहीं होता है , जिसमें माप लिए जाने वाले स्थान परपर्याप्त रक्त संचार नहीं होता या फिर जिसके हृदय की धड़कन बार-बार अनियमितहो जाती हो। मॉनिटर का उपयोग करने या न करने के बारे में अपने चिकित्सक कीसलाह लें।
9. अगर आपकी बाँह में चोट लगी है , तो कफ़ को न लपेटें।
10. उस बाँह पर कफ़ को न लपेटें जिसमें सुई (अंतःशिरा आधान) लगाई गई हो या फिरचिकित्सकीय उपचार के समय रक्त-आधान किया गया हो। ऐसा करने पर जख्म याकोई गंभीर दर्घटना हो सकती है ।
ु11. प्रतिकूल-संक्रमण से बचने के लिए दसू रे संक्रामक व्यक्ति के लिए इस्तेमाल में लाईगई कफ का प्रयोग न करें ।
12. यदि आपको मैस्टे क्टमी के भाग वाली भुजा पर मौजूद कफ के अनुप्रयोग से संबंधितकोई शंका हो, तो अपने चिकित्सक से परामर्श लें।13. मॉनिटर को एनेस्थिसिया के लिए प्रयुक्त होने वाली गैसों जैसी ज्वलनशील गैस केनिकट उपयोग न करें ।  ऐसा करने पर गैस प्रज्ज्वलित हो सकती है और उससेविस्फोट हो सकता है ।
14. मॉनिटर का उपयोग ऑक्सीज़न की अधिक मात्रा वाले स्थानों जैसे हॉस्पिटल काहाइपरबेरिक चैंबर या ऑक्सीजन टैंट में न करें । ऐसा करने पर ऑक्सीजन में चिंगारीउठ सकती है , जिससे आग लग सकती है ।
15. मॉनिटर के निकट मोबाइल फ़ोन का उपयोग न करें , इसके कारण उसमें ख़राबी आसकती है ।
16. अगर आप कार्डियक पेसमेकर का उपयोग करते हैं, तो मॉनिटर का उपयोग करने सेपहले अपने चिकित्सक की सलाह लें।17. ध्यान रखें कि इस उपकरण का उपयोग केवल ब्लड प्रेशर का माप लेने के लिए हीकिया जाना है । इसका उपयोग किसी अन्य कार्य के लिए न करें ।
18. आप केवल हमारे अधिकृत पर्ु जों और सहायक उपकरणों का उपयोग करना सनु िश्चित
करें । ऐसे पुर्जे और सहायक उपकरण, डिवाइस को नुक़सान पहुँचा सकते हैं जो इसकेसाथ उपयोग के लिए स्वीकृत नहीं हैं।19. इस उपकरण का उपयोग नवजात शिशुओं पर न करें ।
20. हो सकता है , कि उस व्यक्ति के ब्लड प्रेशर का माप लेना संभव न हो जो सामान्य अतालता जैसे अलिंद या निलय संबंधी असामान्य धड़कन या अलिंद विकंपन सेपीड़ित है ।

05_CHU304_Hin.indd 3- Hin 3 -6/3/2024 12:33:52 PM

8.
9.

चेतावनीइस डिवाइस में किसी भी संशोधन की अनुमति नहीं है ।चिह्नों की व्याख्या: 
CE अंकन का प्रयोजन EU के सदस्य दे शों में बाज़ार निरीक्षकों कोजानकारी प्रदान करना है ।
: टाइप BF लागू भाग (कफ BF प्रकार का प्रयुक्त भाग है ।): सूखा रखें: चेतावनी: उपयोग करने से पहले निर्दे श पुस्तिका दे खें (पषृ ्ठभूमि के रं ग का चिह्न:नीला। ग्राफिकल संकेत का चिह्न: सफेद।): उपकरण अनुपालन WEEE निर्दे श: निर्माता
: “यूरोपीय प्रतिनिधित्व” के लिए िचह्न

Español

7.

Français

4.
5.
6.

Pyсский

2.
3.

ब्लड प्रेशर मॉनिटर को धपू वाले स्थान पर, अत्यधिक तापमान (55°C से अधिक),निम्न तापमान (- 20°C से कम), अत्यधिक नमी (90% से ऊपर) या बहुत अधिकधूल वाले स्थान पर न रखें।
सुनिश्चित करें कि ब्लड प्रेशर मॉनिटर को ऐसे स्थान पर रखा जाए, जहाँ बच्चों,पालतू जानवरों की पहुँच न हो और/या कीट न हों।ब्लड प्रेशर मॉनिटर या कफ़ को गिरने से बचाएँ और इस पर झटके या कंपन न आने दें ।अगर मॉनिटर को लंबे समय तक के लिए बिना उपयोग के रखना हो, तो उसमें सेबैट्री निकाल दें ।मॉनिटर या कफ़ को खोलने की कोशिश न करें । विनिर्माता द्वारा नामित सेवा केंद्र से पूछें।कफ़ या वायु नलिका को अधिक न मोड़ें।यदि मॉनीटर और कफ बहुत ही गंदे हैं, तो जीवाणुरहित अल्कोहल या उदासीन डिटर्जेंट के द्वारा नमीयुक्त कपड़े से अच्छी तरह से पोछें । इसके बाद उन्हें सूखेकपड़े से पोछें । यदि हॉस्पिटल में आवश्यकता पड़ती है , तो कफ को प्रत्येक सप्ताह
में 2 बार विसंक्रमित करना चाहिए।
थिनर अथवा बेंज़ीन से कभी भी मॉनिटर को साफ़ न करें , ऐसा करने पर मॉनिटरको नुकसान पहुँच सकता है ।कफ़ को साफ़ करने के लिए गीले कपड़े का उपयोग करें । उसे ज़ोर से न रगड़ें,क्योंकि ऐसा करने पर उसमें से वायु लीक होने का खतरा हो सकता है । यह भीध्यान रखें कि उसकी वायु नलिका में पानी न जाए।प्रत्येक 200 बार उपयोग के बाद कफ को साफ करने की अनुशंसा की जाती है ।हिन्दी

1.

Englishरखरखाव संबंधी सावधानियाँ

EMD के लिए EN60601-1-2 के अनुसार डिवाइस का परीक्षणऔर समरूपीकरण किया गया है । इसमें इस बात की कोई गारं टीचेतावनी नहीं है कि यह डिवाइस विद्युत चुंबकीय तरं गों से प्रभावित नहींहोगा। इस डिवाइस का उपयोग उच्च विद्युत चुंबकीय परिवेश  मेंकरने से बचें ।रखरखाव की प्रकृति और आवत्तिृ :इस उत्पाद को इस प्रकार बनाया गया है , जिससे इसे लंबे समय तक उपयोग किया जासके; फिर भी यह सलाह दी जाती है कि इसका उचित कार्य-निष्पादन सुनिश्चित करने केलिए इसकी जाँच हर दो साल में एक बार की जानी चाहिए।

05_CHU304_Hin.indd 4- Hin 4 -6/3/2024 12:33:52 PMप्राकृतिक वातावरण की सुरक्षा:बैटरी और उत्पाद का जीवनकाल समाप्त होने पर इनके निपटान हे तु कृपया राष्ट्रीय और/या स्थानीय पुनर्चक्रण विनियमनों का अनुपालन करके प्राकृतिक वातावरण के संरक्षण मेंसहायता करें ।

WEEE चिह्नEnglishअगर आप इस उत्पाद का निपटान करना चाहते हैं, तो इसे आम घरे लू कूड़ेमें  न मिलाएँ। WEEE निर्दे श (निर्दे श 2012/19/EU) के तहत बने कानन ू केअनसु ार परु ाने विद्युत उपकरणों के लिए एक अलग संग्रह प्रणाली मौजदू है औरयह केवल यरू ोपीय संघ में ही प्रभावी है ।यूरोपीय संघ के बाहर वाले दसू रे देशों में निपटान की जानकारी।

Españolअगर आप पुरानी बैट्री को फेंकना चाहते हैं, तो कृपया अपने स्थानीय प्राधिकारी याविक्रेता से संपर्क करें और उनसे निपटान का सही तरीका पूछें।

बैट्री के चिह्न के लिए नोट।इस चिह्न का उपयोग रासायनिक चिह्न के संयोजन में भी किया जा सकता है । ऐसीस्थिति में रसायन से संबंधित कानूनी निर्दे शों का अनुपालन करना आवश्यक है ।

Françaisसंदर्भ यरू ोपीय मानक:ब्लड प्रेशर मापन डिवाइस EN60601-1, EN1060-1, EN1060-3, EN1060-4 विनियमन से संबंधित है ।कैलिब्रेशन:Pyсский

ब्लड प्रेशर मॉनिटर की जाँच आमतौर पर दो साल में एक बार करवाने की अनुशंसाकी जाते है , जिससे कि आप सुनिश्चित हो सकें कि यह मॉनिटर ठीक से कार्य कररहा है । इसके लिए कृपया वितरक से संपर्क करें ।हिन्दी

नाम :
पता :

फ़ैक्टरी
नाम :
पता :

CITIZEN SYSTEMS JAPAN CO., LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi,
Tokyo 188-8511, Japan
ANDON HEALTH CO., LTD.
No.3 Jinping Street, YaAn Road, Nankai District,Tianjin 300190, Chinaयूरोपीय प्रतिनिधि:
नाम :
पता :

EMERGO EUROPE
Westervoortsedijk 60, 6827 AT Arnhem,
The Netherlands• तकनीकी डेटा के िलए, हमारे यूरोपीय प्रितिनिध से संपकर् करचीन में निर्मित

05_CHU304_Hin.indd 5- Hin 5 -6/3/2024 12:33:52 PMपुर्जों की पहचान
घटकों के नाम
Englishवाय नु लिका

डिस्प्ले

मेमोरी स्विच

स्टार्ट /स्टॉप स्विच

मेटल हूप

वायु नलिका
कनेक्टर

Françaisस्टैंड-बायवायु नलिका सॉकेट

Pyсский

* बैट्री कोष्ठ के लिए पषृ ्ठ 7 दे खें<डिस्प्ले>बैट्री आइकनवायु भरने के लिए तैयार
होने का संकेतक

सिस्टोलिक ब्लडप्रेशर डिस्प्ले
उच्च रक्तचाप
वर्गीकरण संकेतक

05_CHU304_Hin.indd 6

Español<कफ़>हिन्दी

<मुख्य यूनिट>मेमोरी आइकनडायस्टोलिक ब्लडप्रेशर/नाड़ी गति डिस्प्ले

अनियमित हृदय स्पंदन  दिनांक/समय डिस्प्ले
(IHB) का संकेतक** हृदय के अनियमित स्पंदन (IHB) के संकेतक के लिए पषृ ्ठ 11 दे खें।- Hin 6 -6/3/2024 12:33:53 PMबैट्री लगाना
(ब्लड प्रेशर मॉनिटर का आपके द्वारा उपयोग किए जाने से पहले यूनिट के साथ दी गईबैट्री को कोष्ठ में लगाया जाना चाहिए।)English1 बैट्री कोष्ठ का ढक्कन खोलें।

बैट्री के ढक्कन के ऊपरी भाग को दबाते हुए, ढक्कन खोलने के लिए उसे अपनीतरफ खींचें।
बैट्री कोष्ठ

Españolबैट्री का ढक्कन

Français* अगर ब्लड प्रेशर मॉनिटर लंबे समय से उपयोग न किया जा रहा हो, तोबैट्री को कोष्ठ से निकाल दे ना चाहिए।

2 बैट्री को कोष्ठ में रखें।

Pyсский

बैट्री को धनात्मक और ऋणात्मक चिह्नों
औरको ध्यान में रखते हुए लगाएँ।
4 “AA” बैट्री

3 बैट्री के ढक्कन को बंद कर दें ।हिन्दी

बैट्री निकालना• जबआइकन प्रदर्शित हो या डिस्प्ले में कुछ भी प्रदर्शित न हो रहा हो, तो पुरानी
बैट्री के स्थान पर चार नई बैट्री लगा दें । चारों बैट्री एक साथ ही बदली जानी चाहिए।
• बैट्री बदलने के बाद घड़ी समायोजित करें ।• बैट्री बदलने से मेमोरी में सहे जा गया डेटा हटाया नहीं जाता।
• रीचार्जेबल बैट्री का उपयोग न करें ।• ऐल्कलाइन (LR6) और मैंगनीज़ (R6P) बैट्री एक साथ उपयोग न करें ।• यहाँ दिखाई गई बैट्री केवल प्रदर्शन के उद्देश्य से है । बैट्री का जीवनकाल निर्दिष्ट समयकी तुलना में कम हो सकता है ।
• पुरानी बैट्री का निपटान करते समय, अपने दे श/क्षेत्र में लागू होने वाले सरकारी
विनियमों या पर्यावरणीय सार्वजनिक संस्था के नियमों का अनुपालन करें ।

05_CHU304_Hin.indd 7- Hin 7 -6/3/2024 12:33:53 PMस्टार्ट/स्टॉप” स्विच और “मेमोरी” स्विच को एकसाथ दो सेकंड तक दबाए रखें, इससे बीप सनु ाई देगी औरसबसे पहले “माह” संकेतक प्रकाशमान होगा।

3 “मेमोरी” स्विच को दबाकर “माह” समायोजित करें । संख्या मेमोरी” स्विच को दबाएँ।को आगे बढ़ाने के लिए “स्टार्ट /स्टॉप” स्विच दबाएँ।
सेटिगं की पुष्टि करने के लिए ““माह” सेट हो जाता है और “दिन” संकेतक प्रकाशमान होनेलगता है ।*  मेमोरी स्विच को दबाए रखकर, आप संख्या को डिस्प्ले मेंसकते हैं।
तेजी से आगे बढ़ा 4 “

5मेमोरी” स्विच को दबाकर “दिन” समायोजित करें ।मेमोरी” स्विच कोसंख्या को आगे बढ़ाने के लिए “स्टार्ट /स्टॉप”दबाएँ।  सेटिगं की पुष्टि करने के लिए “स्विच दबाएँ। “दिन” सेट हो जाता है और “घंटा” संकेतक
प्रकाशमान होने लगता है ।

Español

2 “Françaisदिनांक बारी-बारी से दिखाएगा।

Pyсский

1	जब आप पहली बार बैट्री लगाएंगे, तो मॉनिटर समय औरहिन्दी

घड़ी को सेट करने से माप सही ढं ग से रिकॉर्ड होती है ।मॉनिटर माप के दिनांक और समय के साथ-साथ माप परिणामों को भी रिकॉर्ड करसकता है । बैट्री लगाने के बाद घड़ी ज़रूर सेट करें ।

Englishसमय का समायोजन

“मेमोरी” स्विच को दबाकर “घंटा” समायोजित करें ।मेमोरी” स्विच कोसंख्या को आगे बढ़ाने के लिए “स्टार्ट /स्टॉप” स्विच
दबाएँ। सेटिगं की पुष्टि करने के लिए “दबाएँ। “घंटा” सेट हो जाता है और “मिनट” संकेतक प्रकाशमान होने लगता है ।

6 “मेमोरी” स्विच को दबाकर “मिनट” समायोजित करें । संख्या मेमोरी” स्विच को दबाएँ।को आगे बढ़ाने के लिए “स्टार्ट /स्टॉप” स्विच दबाएँ।
सेटिगं की पुष्टि करने के लिए “*  घड़ी के समायोजन को पूरा करने के लिए रे डियो आदि परस्टार्ट /स्टॉप” दबाएँ।समय संकेत के साथ “

05_CHU304_Hin.indd 8- Hin 8 -6/3/2024 12:33:54 PM

अपने ब्लड प्रेशर का माप लेना➊	वायु नलिका कनेक्टर को मुख्य यूनिटमें लगाएँ।English• माप के समय वायु नली को न मोड़ें, जिससे
स्फीति त्रुटि हो सकती है , कफ के निरं तर दाब केकारण हानिकारक दर्घटना हो सकती है ।ु➋	कफ़ लगानावाय नु लिका
कनेक्टर

वायु नलिका
सॉकेट

Españolअपना ब्लड प्रेशर कफ़ लगाने से पहले, यह सुनिश्चित करें कि आपने उचित आकारके कफ़ का चयन किया है :कफ़ (मॉडल: SCN-008) इस माप की बाँह में सही बैठता है : 22 सेमी – 30 सेमी।
चौड़ा कफ़ (मॉडल: SCW-009) इस माप की बाँह में सही बैठता है : 22 सेमी – 42सेमी।Français12कफ़ को खोल लें उसके सिरे को धातु हूप तक ले जाएँताकि हुक-और-लूप फैसनर वाला भाग बाहर की तरफरहे।Pyсскийकफ़ को अपनी बाँह के चारों ओर इस तरह से लपेटेंकि वायु नलिका हथेली की तरफ रहे।मेटल हूप

हिन्दी

3: धमनी की स्थिति।कफ़ लगाते समय, कफ़ की स्थिति इस तरह समायोजित करें कि वायु की नलीआपकी बाँह के मध्य भाग में स्थित हो और कफ़ का किनारा आपकी कोहनी से 1सेमी – 2 सेमी ऊपर हो।

05_CHU304_Hin.indd 9* कफ़ को खुली बाँह पर या हल्के अंतरवस्त्रके ऊपर लगाया जाना चाहिए।
* अगर आपने भारी कपड़े पहन रखे हैं, तोकृपया उन्हें निकाल दें ।

1 सेमी – 2 सेमी।

- Hin 9 -6/3/2024 12:33:55 PM

EnglishEspañol* कफ़ को अपनी बाँह पर सावधानीपूर्वक
लगाएँ और ध्यान रखें कि उसकाकसाव सही हो।  यदि कफ़ और बाँहके बीच में आपकी उँ गली आसानी सेचली जाती है तो समझ लें कि उसकाकसाव ठीक है ।अपनी बाँह को ढीला छोड़ दें और अपनी हथेली को ऊपर की ओर रखते हुए थोड़ाखोल कर रखें।

➌ कफ़ को अपने हृदय की समानांतर ऊँचाई पर लगाएँ।
➍ “ स्टार्ट/स्टॉप” स्विच दबाएँ।

• एक बीप सुनाई दे ने के बाद प्रारं भिक प्रदर्शन पर सभी अंक प्रदर्शित होते हैं, जोयह दर्शाता है कि मॉनिटर सामान्य रूप से कार्य कर रहा है ।
• कफ़ में स्वचालित रूप से दबाव बन जाता है ।
* अगर यूनिट को दबाव अपर्याप्त लगता है , तो यह स्वचालित रूप से फिर से दबावदे ती है ।
* यनू िट पिछले माप के आधार पर स्वचालित रूप से दबाव सेट करती है ।

Français6अपनी बाँह को टेबल पर या उसके जैसे किसीअन्य स्थान पर रखें ताकि कफ़ का मध्यभाग आपके हृदय की समानांतर ऊँचाई पररहे।Pyсский

5कफ़ के सिरे को बाहर की ओर खीचें ताकिकफ़ आपकी बाँह के चारों ओर लिपट जाएऔर फिर उसके सिरे को हुक-और-लूप फैसनरमें फंसा दें ।हिन्दी

4*	 तब प्रदर्शित होता है , जब नाड़ी का पता चलता है ।

* माप बंद करने के लिए, “ स्टार्ट /स्टॉप” स्विच दबाएँ। कफ़ से वायु निकल जातीहै और मापन बंद हो जाता है ।

05_CHU304_Hin.indd 10* अगर पिछले उपयोगकर्ता के लिए उच्च दबाव की सेटिगं की गई है , तो स्वचालितरूप से उसी उच्च दबाव सेटिगं का उपयोग वर्तमान माप के लिए किया जाएगा।

* अगर दबाव का मान 260 mmHg से अधिक हो जाता है और दबाव असामान्य लगता है या अगर आप मापने की प्रक्रिया को रोकना चाहते हैं, तो“ स्टार्ट /स्टॉप” स्विच को दोबारा दबाएँ। कफ़ की वायु निकल जाती है औरपॉवर बंद हो जाता है ।

- Hin 10 -6/3/2024 12:33:55 PM➎ माप के परिणाम प्रदर्शित होते हैं।English• माप पूरा हो जाने के बाद, कफ़ की वायुनिकल जाती है और माप के परिणाम
(सिस्टोलिक/डायस्टोलिक ब्लड प्रेशर मान,नाड़ी गति) प्रदर्शित हो जाते हैं।* सिस्टोलिक/डायस्टोलिक ब्लड प्रेशर मान औरनाड़ी गति बारी-बारी से प्रदर्शित होते हैं।* माप परिणाम में कोई त्रुटि नहीं होनेपर, उन्हें स्वचालित रूप से संग्रहीत करलिया जाता है ।उच्च रक्तचाप वर्गीकरण संकेतक

Español➏ मापन समाप्त करना।

• मॉनिटर को बंद करने के लिए “ स्टार्ट /स्टॉप” स्विच दबाएँ।
• ब्लड प्रेशर का माप लेने के बाद, “ स्टार्ट /स्टॉप” स्विच दबाने से मॉनिटर बंद होजाएगा। हालाँकि, अगर आप स्विच नहीं दबाते हैं, तब भी 1 मिनट बाद, मॉनिटर कास्वतः बंद फ़ं क्शन इसे स्वचालित रूप से बंद कर दे गा।

Françaisअनियमित हृदय स्पंदन का संकेतकअगर माप के दौरान, अनियमित हृदय स्पंदन का पता चलता है , तो माप केपरू ा होने के बाद, अनियमित हृदय स्पंदन (IHB) संकेतक प्रदर्शित होता है ।

Pyсский

2 स्थितियाँ ऐसी हैं, जिनमें IHB का संकेत प्रदर्शित होता है :1) नाड़ी स्पंदन की अवधि का भिन्नता गुणांक > 25%
0.14s और ऐसी नाड़ी की संख्या,2) सन्निकट नाड़ी स्पंदन अवधि का अंतरनाड़ी की कुल संख्या का अधिकतम 53 प्रतिशत होती है ।
• अगर मापन के दौरान, आपके हृदय स्पंदन में अधिक उतार-चढ़ाव होता रहता है , तो मापन मेंत्रुटि हो सकती है । अगर अनियमित हृदय स्पंदन (IHB) का संकेतक प्रदर्शित होता है , तो अपनेब्लड प्रेशर का दोबारा माप लें।सावधानी
हिन्दीअगर अनियमित हृदय स्पंदन (IHB) का संकेतक बार-बार प्रदर्शित
होता है , तो आपको अपने स्वास्थ्य की स्थिति के बारे में चिकित्सकसे परामर्श करने की सलाह दी जाती है ।उच्च रक्तचाप वर्गीकरण संकेतक

मापा गया ब्लड प्रेशर मान WHO के मानकों के अनुसार प्रदर्शित किया जाता है ।पषृ ्ठ 2 दे खें)(05_CHU304_Hin.indd 11सिस्टोलिक (mmHg)
180
160
140
130
120

श्रेणी 3 उच्च रक्तचाप (गंभीर)श्रेणी 2 उच्च रक्तचाप (मध्यम)श्रेणी 1 उच्च रक्तचाप (हल्का)उच्च-सामान्य
सामान्य
आदर्श
80

85

90 100- Hin 11 -110डायस्टोलिक
(mmHg)

6/3/2024 12:33:56 PM

■	लेटी हुई मुद्रा में माप लेना* कफ़ पर दबाव पड़ने से बचाने केलिए अपने आप को शिथिल रखें।
■ आप अपने ब्लड प्रेशर को मापने के लिए अपने दाईं बाँह का उपयोग कर सकते हैं।बाईं बाँह के बजाय दाईं बाँह (या इसके विपरीत) का उपयोग करने पर, ब्लड प्रेशर के मान में 10 mmHg से अधिक का अंतर होने की संभावना रहती है । अपने ब्लड प्रेशर का मापप्रतिदिन उसी बाँह पर लें।मेमोरी में संग्रहीत डेटा निकालना

मेमोरी में 99 मापन परिणाम संग्रहीत किए जा सकते हैं।आपके दै निक स्वास्थ्य को प्रबंधित करने मेंसहायता के लिए औसत मान की स्वचालित रूपसे गणना की जाती है ।

1

Español■ बैठकर माप लेना* फर्श पर पैरों को फै लाकर आराम से बैठें, और एक पैर को दसू रे पैर पर न रखें।
➀ अपनी बाँह को टे बल या वैसी ही किसी सतह के ऊपर आगे की ओर करके रखें।
➁ बाँह को अपने हृदय के समानांतर रखना आवश्यक है।➂ टे बल के ऊपर अपने हाथ को थोड़ा-सा खोलें, जिससे आपकीहथेली ऊपर की तरफ रहे और आपकी अंगलु ियाँ ढीली रहें ।➃ माप लेते समय अपने शरीर को न तो हिलाएँ और न ही कुछ बोलें।

Françaisआपका ब्लड प्रेशर, आपकी शारीरिक स्थिति, दिन के समय और अन्य कारकों की श्रेणी केअनसु ार बदलता रहता है । इसलिए, आपको अपना ब्लड प्रेशर प्रतिदिन एक ही समय औरएक जैसी शारीरिक स्थिति में मापना चाहिए।

Pyсский

सटीक माप प्राप्त करने की यकु ्तियाँ

हिन्दीयदि निर्धारित रक्त दाब (सिस्टोलिक या डायस्टोलिक) विनिर्दे श भाग में निर्दिष्ट मूल्यांकित सीमा के बाहर हो,तो मॉनीटर विलंब किए बिना LCD पर तकनीकी अलार्म के रूप में ‘HI’ या ‘Lo’ प्रदर्शित करे गा। इस स्थिति में ,आपको चिकित्सक से परामर्श लेना चाहिए या यह जांचना चाहिए कि आपकी कार्य-प्रणाली निर्दे शों का उल्लंघन करतीहै या नहीं। तकनीकी अलार्म की स्थिति (मूल्यांकित सीमा के बाहर) फैक्ट्री में प्रीसेट होती है और इसे समायोजितया निष्क्रिय नहीं किया जा सकता है । इस अलार्म स्थिति को IEC 60601-1-8 के अनुसार न्यून प्राथमिकता के रूपमें निर्दिष्ट किया जाता है । तकनीकी अलार्म बिना किसी कंु डी के होता है और इसे रीसेट करने की आवश्यकता नहींपड़ती है । LCD पर प्रदर्शित होने वाले संकेत लगभग 8 सेकंड के बाद स्वचालित रूप से गायब हो जाएगा।Englishतकनीकी अलार्म का विवरण

मेमोरी स्विच

मेमोरी स्विच को एक बार दबाएँ।
मेमोरी स्विच को दबाए जाने पर प्रदर्शन का उदाहरण।
*	अब LCD पर वर्तमान बैंक में संग्रहीत परिणामों की मात्रा प्रदर्शित हो जाती है और, डिस्प्ले के निचले भाग पर प्रदर्शित औसत मान का संकेत दे ता है । उसके बाद,पिछले तीन मापों के आधार पर औसत मान प्रदर्शित होता है ।

05_CHU304_Hin.indd 12- Hin 12 -6/3/2024 12:33:57 PM

2अगर आप
मेमोरी स्विच को दोबारा दबाते हैं, तो पूर्व के मापनों के डेटा प्रदर्शित
होते हैं।
जितनी बार आप स्विच दबाते जाते हैं, सबसे हाल के माप डेटा से लेकर सबसे पुराने डेटा केक्रम में माप डेटा प्रदर्शित होता है ।*मेमोरी स्विच को दबाने पर प्रदर्शन का उदाहरण।EnglishEspañolडेटा संख्या मापपरिणाम से पहलेप्रदर्शित की जाती है ।

Français• मेमोरी में संग्रहीत मान, मापन क्रम के अनसु ार होता है । उदाहरण के लिए, जब मेमोरी में डेटाके 99 सेट संग्रहीत होते हैं तो डेटा संख्या 1 नवीनतम डेटा दर्शाती है ।डेटा संख्या: 1, 2, 3, ...
... 98, 99ÇÇनवीनतम डेटासबसे पुराना डेटाPyсский

1डेटा हटाने का तरीका

किसी भी परिणाम (पिछली तीन मापों की औसत रीडिंग को छोड़कर) के प्रदर्शित होनेमेमोरी स्विच को 3 सेकंड या अधिक समय तक दबाए रखें।
परमेमोरी में संग्रहीत सभी डेटा हटा दिए जाएँगे।

3 सेकंड तकदबाए रखें

हिन्दीहटा दिया गयामेमोरी से संपूर्ण डेटा हटा दिएजाने की जानकारी दे ता हुआएक संदेश प्रदर्शित होता है ।"ब्लड प्रेशर" के बारे में जानकारी

ब्लड प्रेशर क्या है ?

(सिस्टोलिक)(डायस्टोलिक)धमनीहृदय एक प्रकार का पंप है ,धमनीजो आपके शरीर में रक्त काबायाँ अलिंद
बायाँ अलिंद
परिसंचरण करता है । हृदय सेरक्त नियत दबाव से धमनियों
बायाँधमनी की दीवार मेंबायाँमें पंप किया जाता है । यह दबाव
निलय
निलय
सामान्य स्थिति धमनी की दीवार
पुनर्स्थापित हो गई है ।धमनीय ब्लड प्रेशर कहलाता है औरफैलती है ।सामान्य शब्दों में , इसे ही ब्लडसंकुचन
प्रसार
प्रेशर कहा जाता है । ब्लड प्रेशर
हृदय से रक्त को पंप किए जाते समय प्रयुक्त होने वाले सिस्टोलिक प्रेशर और रक्त कोवापस हृदय में लाते समय प्रयुक्त होने वाले डायस्टोलिक प्रेशर सहित, अनेक प्रकार केदबावों द्वारा इंगित किया जाता है ।

05_CHU304_Hin.indd 13- Hin 13 -6/3/2024 12:33:57 PM

140
120
100
80
60
40

सोते समय
12 13 14 15 16 17 18 19 20 21 22 23 24 1

2

3

4

5

6

7

8

डायस्टोलिक
ब्लड प्रेशर

(पूरे दिन पाँच मिनट के अंतरालपर मापे गए ब्लड प्रेशर के मान)● Bevan AT, Honour
AJ, Stott FH. Clin Sci
1969;36:329–44.

9 10 11दिन का समयब्लड प्रेशर के बारे में प्रश्नोत्तर
प्रश्नघर पर मापा गया ब्लड प्रेशर माप, डॉक्टर द्वारा या हॉस्पिटल में मापे गए ब्लड प्रेशरसे भिन्न क्यों होता है ?

उत्तर आपका ब्लड प्रेशर व्यायाम, परिवेशी तापमान या आपकी मनोदशा के कारण भिन्न होसकता है । जब आपके ब्लड प्रेशर का माप कोई डॉक्टर या नर्स लेती है , तो चिंता यातनाव के कारण प्रेशर का मान सामान्य से 10 से 20 mmHg तक अधिक हो सकता है ।आपके ब्लड प्रेशर को प्रभावित करने वाले कारकों के बारे में  जानने से आपके स्वास्थ्य की बेहतर निगरानी करने में सहायता मिलती है ।प्रश्न

ब्लड प्रेशर का माप हर बार अलग क्यों आता है ?

Españolसिस्टोलिक ब्लड प्रेशर

160

Françaisब्लड प्रेशर के अंक ( mmHg )[ब्लड प्रेशर में दिनभर की भिन्नताओं का उदाहरण]Pyсский

आपका ब्लड प्रेशर, आपकी आयु, लिंग और अन्य कई कारकों के अनुसार बदलता
रहता है । इसके, आपके पूरे दिन के बायोरिद्म और आपकी शारीरिक स्थिति से, शारीरिक
गतिविधियों, मानसिक गतिविधियों, तनाव के स्तर और हवा के तापमान से भी प्रभावित
होने की संभावना रहती है । एक स्वस्थ व्यक्ति का ब्लड प्रेशर आम तौर पर दिनभर
बदलता रहता है ।

Englishआपका ब्लड प्रेशर हमेशा बदलता रहता है ।प्रश्न

प्रेशर प्रत्येक हृदय स्पंदन पर भिन्न होता है । हमें यह लग सकता है कि हमारा ब्लडप्रेशर स्थिर है , लेकिन आपके द्वारा बार-बार माप लेने पर यह अलग-अलग आता है ।ब्लड प्रेशर, दिन के समय जैसे सुबह और दोपहर, मौसम और वायुमंडलीय ताप केप्रति संवेदनशील होता है । इसके अतिरिक्त, ब्लड प्रेशर, मानसिक तनाव या भावनात्मक
उतार-चढ़ाव से आसानी से प्रभावित हो जाता है । आपके तनावग्रस्त होने पर यह बढ़ जाता है या आपके चिंतामुक्त होने पर यह कम हो जाता है ।हिन्दी

उत्तर आपका ब्लड प्रेशर स्वचालित तंत्रिका प्रकार्य द्वारा समायोजित किया जाता है । ब्लडघर पर ब्लड प्रेशर का माप लेने के क्या लाभ हैं?

उत्तर घर पर ब्लड प्रेशर का माप लेने से आपको भरोसमंद डेटा मिलता है क्योंकि घर पर मापन,05_CHU304_Hin.indd 14

स्थिर और शांत अवस्था में किया जा सकता है । ब्लड प्रेशर के घर पर लिए गए माप कोचिकित्सक बहुत महत्व दे ते हैं। आपको अपने ब्लड प्रेशर का माप प्रतिदिन एक ही समय परलेना चाहिए और अपने ब्लड प्रेशर के परिवर्तन दर्ज करने चाहिए।

अपने ब्लड प्रेशर रीडिंग की व्याख्या और उसका उचित उपचार निर्धारित करने केलिए हमेशा अपने चिकित्सक से परामर्श लें।- Hin 14 -6/3/2024 12:33:57 PMउपकरण की मरम्मत या जाँच के लिए अनरु ोध करने से परू ्वअपने ब्लड प्रेशर मॉनिटर को जाँच और मरम्मत के लिए भेजने से पहले निम्न की अच्छीतरह जाँच करना सुनिश्चित करें ।

Englishसमस्या निवारण (1)समस्या

Español

LCD डिस्प्ले असामान्य परिणाम
दिखाता हैसंभावित कारणकफ़ की स्थिति सही नहीं थी या इसे ठीक सेकसा नहीं गया थापरीक्षण के दौरान शरीर की मुद्रा सहीनहीं थी

Françaisसमस्या

Pyсский

LCD बैट्री कम होने काचिह्न दिखाती हैनिर्दे श और पुनः परीक्षण के "माप के दौरान शरीर की मुद्रा" से संबंधित अनुभागों कोफिर से दे खें।परीक्षण के दौरान बोलना, हाथ या शरीरपरीक्षण के दौरान शांत रहकर और बोलेका हिलना, क्रोधित, उत्तेजित या परे शान या हिले बिना पुनः परीक्षण करेंहोना।गंभीर अतालता से पीड़ित लोगों के लिएइस इलेक्ट्रॉनिक रक्तदाबमापी का उपयोग
अनुचित होता है ।अनियमित हृदय स्पंदन
(अतालता)समस्या निवारण (2)समाधानकफ़ को सही तरीके से लगाएँ और पुनःप्रयास करेंसंभावित कारणसमाधानकम बैट्री

बैट्री बदलें

हिन्दी

LCD “Er 0” दिखातीमाप से पहले दबाव प्रणाली अस्थिर हैहैLCD “Er 1” दिखाती
सिस्टोलिक प्रेशर का पता लगाने मेंहिलें नहीं और पुनः प्रयास करें ।हैविफल रहा
LCD “Er 2” दिखाती डायस्टोलिक प्रेशर का पता लगाने मेंहैविफल रहा
LCD “Er 3” दिखाती न्यूमेटिक प्रणाली अवरुद्ध है या प्रसारित
हैहोने के दौरान कफ़ का कसाव अधिक है कफ़ को सही तरीके से लगाएँ औरपुनः प्रयास करेंLCD “Er 4” दिखाती न्यूमेटिक प्रणाली में रिसाव है या प्रसारित
हैहोने के दौरान कफ़ का कसाव कम है
LCD “Er 5” दिखातीकफ़ का दबावहै
300 mmHg से अधिक है
LCD “Er 6” दिखाती
3 मिनट से अधिक समय तक 15
है
mmHg तक का कफ़ दबावपाँच मिनट के बाद फिर से मापें ।अगर मॉनिटर अभी भी असामान्य LCD “Er 7” दिखातीEEPROM पहुँच में  त्रुटि
है , तो स्थानीय वितरक या फैक्ट्री सेहैसंपर्क करें ।
LCD “Er 8” दिखाती
डिवाइस पैरामीटर जाँच में  त्रुटिहै
LCD “Er A”
प्रेशर सेंसर पैरामीटर त्रुटि
दिखाती हैआपके द्वारा बटन दबाए जाने या बैटरी
लगाए जाने पर कोईप्रतिक्रिया नहीं होना।गलत संचालन या प्रबल विद्युत
चुंबकीय व्यवधान

बैट्री पाँच मिनट के लिए बाहर निकाललें और फिर सभी बैट्री फिर से लगाएँ।

* अगर आप इस उत्पाद का निपटान करना चाहते हैं, तो इसे आम घरे लू कूड़े में  न मिलाएँ। पुराने
विद्युत उपकरणों के लिए, कानून के अनुसार एक अलग संग्रह प्रणाली है ।

05_CHU304_Hin.indd 15- Hin 15 -6/3/2024 12:33:58 PMविनिर्दे शनियत वोल्टेज:6V DC  

विद्युत आपूर्ति:4 X 1.5V
SIZE AA बैटरियाँ (LR6)लगभग 800 बार (ऐल्कलाइन)लगभग 1 मिनट (सक्रिय होने के बाद)138 (W) x 54 (H) x 95 (D) मिमीलगभग 148 × 510 मिमीलगभग 163 × 580 मिमीइकाई: लगभग 207 ग्रा बैट्री के बिनाकफ़: लगभग 128 ग्रा., चौड़ा कफ़: लगभग 134 ग्रा.
10°C – 40°C

आइकनसंकुचन:वजन:संचालन की स्थितियाँ:भंडारण की स्थिति:विद्युत आघात से सुरक्षा:(: दिष्ट धारा)विद्युतचुंबकीय त्वरित निकास वाल्व

बैट्री की अवधि:स्वतः पॉवर ऑफ़ फ़ं क्शन:कफ़ का आकार:Englishस्वचालित गति संकुचन प्रणाली।

निकास:

मुख्य यूनिट आयाम:Españolस्विच:
प्रसार:: माप आइकन: वायु भरने के लिए तैयार होने का संकेतक
: बैट्री आइकन: अनियमित हृदय स्पंदन (IHB) का संकेतक
: मेमोरी आइकन: उच्च रक्तचाप वर्गीकरण संकेतक
2 (स्टार्ट /स्टॉप, मेमोरी)आंतरिक पंप द्वारा स्वतः प्रसार की दर

LCD प्रदर्शित करती है :Françaisशुद्धता:

दाब

नाड़ी
दाब
नाड़ी
दाब
नाड़ी

कफ़
चौड़ा कफ़तापमान
आर्द्रता
तापमान
आर्द्रता

सुरक्षा की श्रेणी

Pyсскийमाप सीमा:CHU304, CHU305
दोलायमान विधिडिजिटल डिस्प्ले प्रकार
बाँह का ऊपरी भागमुलायम कफ़
22 सेमी – 30 सेमी (कफ़), 22 सेमी – 42 सेमी (चौड़े कफ़)0 – 300 mmHg
सिस्टोलिक: 60 – 260 mmHg
डायस्टोलिक: 40 – 199 mmHg
40 से 180 पल्स/मिनट
±3 mmHgरीडिंग का ±5%
3 अंक
3 अंक

90%RH
-20°C – 55°C

हिन्दी

मॉडल नंबर:मापन प्रणाली:डिस्प्ले
मापन स्थानीयकरण
कफ़:कफ़ परिधि सीमा:कफ़ दबाव:90%RHआंतरिक पॉवर यूनिट

संचालन मोड:मेमोरी:सेवाकाल:कफ़ का जीवनकाल: टाइप BF लागू भागसतत संचालन
1 × 99 रीडिंग, अंतिम 3 रीडिंग का औसत
10,000 माप या 3 वर्षलगभग 1000 बारसहायक उपकरण:सेट में एक कफ़, मॉनिटर के लिए 4 AA बैट्री (LR6), निर्दे श पुस्तिका शामिल हैंपानी के प्रवेश से सुरक्षा:वैकल्पिक सहायक उपकरण:* इस डिवाइस के लिए अनुप्रयुक्त हिस्सा कफ़ है ।

IPX0

कफ़ 22 सेमी – 30 सेमी, चौड़े कफ़ 22 सेमी – 42 सेमी* बैरोमीटर दबाव की सीमा 800 hPa से 1050 hPa* यह डिवाइस नीचे दिए गए मानकों के अनुरूप है : IEC 60601-1/EN 60601-1 (चिकित्सा विद्युत उपकरण - भाग 1),
IEC 60601-1-2/EN 60601-1-2 (चिकित्सा विद्युत उपकरण - भाग 1-2), EN 1060-3 (गैर-आक्रामक स्फ़िग्मोमनोमीटर्स - भाग 3), IEC 80601–2–30 (चिकित्सा विद्युत उपकरण - भाग 2-30)

05_CHU304_Hin.indd 16- Hin 16 -6/3/2024 12:33:58 PM

00_CHU304_305_2401.indb 1

MEMO

8/3/2024 9:21:36 AM

00_CHU304_305_2401.indb 2

MEMO

8/3/2024 9:21:36 AM

‫المواصفات‬‫الوزن‪:‬‬‫ظروف التشغيل‪:‬‬‫ظروف التخزين‪:‬‬‫الحماية من الصدمة الكهربية‪:‬‬‫درجة الحماية‬‫وضع التشغيل‪:‬‬‫الذاكرة‪::‬‬‫مدة الخدمة‪:‬‬‫عمر الرباط‪:‬‬‫الوقاية من دخول الماء‪:‬‬‫الملحقات‪:‬‬‫درجة الحرارة‬‫الرطوبة‬‫درجة الحرارة‬‫الرطوبة‬‫‪IPX0‬‬‫مجموعة تتضمن رباط البالغين‪ ،‬و‪ 4‬بطاريات )‪ AA (LR6‬لجهاز‬‫المراقبة‪ ،‬ودليل التعليمات‪.‬‬‫الطوق ‪ 22‬سم – ‪ 30‬سم‪ ،‬طوق ذو نطاق عريض ‪ 22‬سم – ‪ 42‬سم‬‫الملحقات االختيارية‪:‬‬‫* الجزء العملي لهذا الجهاز هو الرباط‪.‬‬‫* نطاق بالضغط الجوي يتراوح من ‪ ٨٠٠‬هيكتوباسكال إلى ‪ ١٠٥٠‬هيكتوباسكال‬‫* يتوافق هذا الجهاز مع المعايير التالية‪( IEC 60601-1/EN 60601-1 :‬األجهزة الكهربائية الطبية – الفقرة ‪IEC 60601-1-2/ ،)1‬‬‫‪( EN 60601-1-2‬األجهزة الكهربائية الطبية – الفقرة ‪( EN 1060-3 ،)1-2‬أجهزة قياس ضغط الدم اإللكترونية غير الباضعة ‪ -‬الفقرة ‪،)3‬‬‫‪( IEC 80601–2–30‬المعدات الطبية الكهربائية – الفقرة ‪)2–30‬‬‫‪ -‬العربية ‪- 16‬‬‫‪6/3/2024 12:29:21 PM‬‬‫‪हिन्दी‬‬‫حجم الرباط‪:‬‬‫الطوق‬‫طوق ذو نطاق عريض‬‫‪Pyсский‬‬‫المفاتيح‪:‬‬‫النفخ‪:‬‬‫تفريغ الهواء‪:‬‬‫الفولتية المقدرة‪:‬‬‫التفريغ‪:‬‬‫مصدر اإلمداد بالطاقة‪:‬‬‫مدة بقاء البطاريات‪:‬‬‫وظيفة اإليقاف التلقائي‪:‬‬‫أبعاد الوحدة الرئيسية‪:‬‬‫األيقونات‬‫‪Français‬‬‫شاشة ‪ LCD‬تعرض‪:‬‬‫النبض‬‫الضغط‬‫النبض‬‫الضغط‬‫النبض‬‫‪Español‬‬‫الدقة‪:‬‬‫الضغط‬‫طريقة قياس الذبذبات‬‫نوع الشاشة الرقمي‬‫أعلى الذراع‬‫رباط ناعم‬‫‪ 22‬سم – ‪ 30‬سم (الطوق)‪ 22 ,‬سم – ‪ 42‬سم (طوق النطاق العريض)‬‫‪ 300 – 0‬مم زئبق‬‫االنقباض‪ 260 – 60 :‬ملي متر زئبق‬‫االنبساط‪ 199 – 40 :‬ملي متر زئبق‬‫‪ 40‬إلى ‪ 180‬نبضة‪/‬دقيقة‬‫‪ 3±‬مم زئبق‬‫‪ %5±‬من القراءة‬‫‪ 3‬أرقام‬‫‪ 3‬أرقام‬‫‪ :‬أيقونة القياس‬‫‪ :‬جاهز لمؤشر النفخ‬‫‪ :‬أيقونة البطارية‬‫‪ :‬مؤشر عدم انتظام ضربات القلب (‪)IHB‬‬‫‪:‬أيقونة الذاكرة‬‫‪ :‬مؤشر تصنيف ارتفاع ضغط الدم‬‫‪"( 2‬بدء‪/‬إيقاف (‪ ،")START/STOP‬و"الذاكرة (‪)")M‬‬‫نفخ تلقائي بواسطة مضخة داخلية‬‫نظام تفريغ سريع تلقائي‬‫‪ :‬تيار مستمر)‬‫(‬‫‪ 6‬فولت تيار مستمر‬‫صمام تفريغ كهرومغناطيسي سريع‬‫مقاس ‪)LR6( AA‬‬‫بطاريات ‪ 5.1 × 4‬فولت‬‫حوالي ‪ 800‬مرة تقريبًا (قاعدي)‬‫حوالي دقيقة واحدة (بعد التنشيط)‬‫‪( 138‬عرض) × ‪( 54‬ارتفاع) × ‪( 95‬عمق) مم‬‫ما يقرب من ‪ 510 × 148‬مللي متر‬‫ما يقرب من ‪ 580 × 163‬مللي متر‬‫الوحدة‪ :‬حوالي ‪ 207‬جم بدون البطاريات‬‫الطوق‪ :‬تقريبًا ‪ 128‬جرام‪ ،‬طوق ذو نطاق عريض‪ 134 :‬جرام تقريبًا‬‫‪ °10‬مئوية – ‪°40‬مئوية‬‫‪%90‬من الرطوبة النسبية‬‫‪ °20‬مئوية – ‪°55‬مئوية‬‫‪%90‬من الرطوبة النسبية‬‫وحدة طاقة داخلية‬‫الجزء التطبيقي طراز ‪BF‬‬‫تشغيل مستمر‬‫‪ 99 × 1‬قراءة‪ ،‬القيمة المتوسطة آلخر ‪ 3‬قراءات‬‫‪10000‬قياس أو ‪ 3‬سنوات‬‫حوالي ‪ 1000‬مرة تقريبًا‬‫‪English‬‬‫رقم الطراز‪:‬‬‫نظام القياس‪:‬‬‫الشاشة‬‫موضع القياس‬‫الرباط‪:‬‬‫نطاق محيط الذراع‪:‬‬‫ضغط الطوق‪:‬‬‫نطاق القياس‪:‬‬‫‪CHU305 ,CHU304‬‬‫‪06_CHU304_2401_Ara.indd 16‬‬‫قبل طلب اإلصالح أو االختبار‬‫تأكد من فحص ما يلي قبل إرسال جهاز مراقبة ضغط الدم لالختبار أو اإلصالح‪.‬‬‫استكشاف المشكالت وحلها (‪)1‬‬‫‪English‬‬‫األسباب المحتملة‬‫المشكلة‬‫الحل‬‫‪Español‬‬‫موضع الطوق غير صحيح أو لم يتم إحكام ربطه استخدم الطوق بشكل صحيح وحاول مرة‬‫أخرى‪.‬‬‫قم بمراجعة أقسام "وضعية الجسم أثناء‬‫وضعية الجسم غير صحيحة أثناء االختبار القياس" للحصول على التعليمات وإعادة‬‫االختبار‪.‬‬‫شاشة ‪ LCD‬تعرض نتيجة‬‫قم بإعادة االختبار عندما تكون هاد ًئا وبدون‬‫التحدث أو حركة الذراع أو الجسم‬‫غير طبيعية‬‫التحدث أو الحركة أثناء االختبار‪.‬‬‫والغضب واإلثارة أثناء االختبار‬‫‪Français‬‬‫ضربات القلب غير المنتظمة (اضطراب‬‫النظم)‬‫ليس من المناسب لألشخاص الذين يعانون‬‫من اضطراب النظم الستخدام مانومتر الدم‬‫اإللكتروني‪.‬‬‫األسباب المحتملة‬‫الحل‬‫استكشاف المشكالت وحلها (‪)2‬‬‫المشكلة‬‫‪Pyсский‬‬‫‪हिन्दी‬‬‫شاشة ‪ LCD‬تعرض رمز‬‫التغيير في البطارية‪.‬‬‫البطارية منخفضة‬‫البطارية ضعيفة‬‫شاشة ‪ LCD‬تعرض‬‫نظام الضغط غير مستقر قبل القياس‬‫"‪"Er 0‬‬‫شاشة ‪ LCD‬تعرض‬‫فشل اكتشاف انقباض القلب‬‫ال تقم بتحريك وحاول مرة أخرى‪.‬‬‫"‪"Er 1‬‬‫شاشة ‪ LCD‬تعرض‬‫فشل اكتشاف انبساط القلب‬‫"‪"Er 2‬‬‫شاشة ‪ LCD‬تعرض‬‫تم قفل نظام التنفس الهوائي أو أن الطوق‬‫تم إحكام ربطه أثناء النفخ‬‫"‪"Er 3‬‬‫استخدم الطوق بشكل صحيح وحاول مرة‬‫أخرى‪.‬‬‫شاشة ‪ LCD‬تعرض‬‫تسرب نظام التنفس الهوائي أو أن الطوق‬‫مرتخي بشدة أثناء النفخ‬‫"‪"Er 4‬‬‫ضغط الطوق أعلى من ‪ 300‬ملي متر‬‫شاشة ‪ LCD‬تعرض‬‫زئبق‬‫"‪"Er 5‬‬‫أكثر من ‪ 3‬دقائق وضغط الطوق أعلى من‬‫شاشة ‪ LCD‬تعرض‬‫‪ 15‬ملي متر زئبق‬‫"‪"Er 6‬‬‫القياس مرة أخرى بعد خمس دقائق‪ .‬إذا‬‫شاشة ‪ LCD‬تعرض‬‫كانت الشاشة مازالت غير طبيعية‪ ،‬يرجى‬‫خطأ الوصول إلى ‪EEPROM‬‬‫"‪"Er 7‬‬‫االتصال بالموزع المحلي أو المصنع‪.‬‬‫شاشة ‪ LCD‬تعرض‬‫خطأ في التحقق من معلمة الجهاز‬‫"‪"Er 8‬‬‫شاشة ‪ LCD‬تعرض‬‫خطأ في معلمة استشعار الضغط‬‫"‪"Er A‬‬‫ال توجد استجابة عندما‬‫قم بإخراج البطاريات لمدة خمس دقائق‪ ،‬ثم‬‫التشغيل غير الصحيح أو تداخل‬‫تقوم بالضغط على الزر‬‫قم بإعادة تركيب البطاريات‪.‬‬‫كهرومغناطيسي قوي‬‫أو تحميل البطارية‪.‬‬‫* إذا كنت ترغب في التخلص من هذا المنتج‪ ،‬فال يجوز إلقاؤه مع القمامة المنزلية المعتادة‪ .‬توجد أنظمة تجميع منفصلة للمنتجات‬‫اإللكترونية المستهلكة بموجب التشريعات القانونية‪.‬‬‫‪ -‬العربية ‪- 15‬‬‫‪6/3/2024 12:29:21 PM‬‬‫‪06_CHU304_2401_Ara.indd 15‬‬‫يتغير ضغط الدم لديك طوال الوقت‪.‬‬‫قيمة ضغط الدم (مم زئبق)‬‫‪80‬‬‫‪80‬‬‫ضغط دم‬‫انبساطي‬‫‪9 10 11‬‬‫‪60‬‬‫‪60‬‬‫‪40‬‬‫‪40‬‬‫أثناء النوم‬
‫‪8‬‬

‫‪7‬‬

‫‪6‬‬

‫‪5‬‬

‫‪6 7 8 9 10 11‬‬

‫‪4‬‬

‫‪3‬‬

‫‪4 5‬‬‫‪2‬‬‫‪12 13 14 15 16 17 18 19 20 21 22 23 24 1‬‬‫‪12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3‬‬‫التوقيت‬‫س و ج حول ضغط الدم‬

‫س‬
‫ج‬

‫لماذا يختلف ضغط الدم المُقاس في المنزل عن ضغط الدم المُقاس بواسطة الطبيب في المستشفى؟‬‫قد يختلف ضغط الدم لديك بسبب التمارين أو درجة الحرارة المحيطة أو الحالة الذهنية‪ .‬فعندما تقوم بقياس ضغط الدم‬‫لديك بواسطة طبيب أو ممرضة‪ ،‬ربما تكون قيمة الضغط أعلى بمقدار ‪ 10‬إلى ‪ 20‬مم زئبق عن الطبيعي نظرً ا للقلق و‪/‬‬‫أو اإلجهاد‪ .‬وتساعدك معرفة ما يؤثر على ضغط الدم على مراقبة صحتك بشكل أفضل‪.‬‬‫س‬‫ج ضغط الدم لدينا مضبوط بوظيفة األعصاب التلقائية‪ .‬ويختلف ضغط الدم مع كل ضربة قلب‪ .‬وربما نعتقد أن ضغط الدم‬‫لماذا يختلف ضغط الدم بأخذ القياسات؟‬

‫س‬
‫ج‬

‫لدينا ثابت‪ ،‬إال أنه يتغير إذا أخذت قياسات بصفة متكررة‪ .‬فضغط الدم يتأثر بتوقيت اليوم كالصباح والظهيرة‪ ،‬والفصول‪،‬‬‫ودرجة حرارة الجو‪ .‬وعالوة على ذلك‪ ،‬يتأثر ضغط الدم باإلجهاد الذهني أو الحاالت العاطفية المختلفة بسهولة‪ .‬فهو يميل‬‫لالرتفاع عندما تكون متوترً ا‪ ،‬ولالنخفاض عندما تكون مسترخيًا‪.‬‬‫ما الفائدة من قياس ضغط الدم لدي في المنزل؟‬‫قياسات ضغط الدم المأخوذة في البيت تعطي بيانات موثوقة‪ ،‬إذ أنها تؤخذ عندما تكون في حالة مستقرة ومسترخية‪.‬‬‫ويولي األطباء اهتمامًا كبيرً ا لقياسات ضغط الدم المأخوذة في المنزل‪ .‬وينبغي عليك قياس ضغط الدم لديك في نفس‬‫الوقت من كل يوم‪ ،‬وتسجيل االختالفات في ضغط الدم لديك‪.‬‬‫استشر دائمًا طبيبك حول تفسير قراءات ضغط الدم لديك وتحديد العالج المناسب‪.‬‬‫‪ -‬العربية ‪- 14‬‬‫‪6/3/2024 12:29:21 PM‬‬‫‪हिन्दी‬‬‫‪100‬‬‫‪100‬‬‫‪Pyсский‬‬‫‪● Bevan AT, Honour AJ, Stott FH.‬‬‫‪Clin Sci 1969;36:329–44‬‬‫‪120‬‬‫‪120‬‬‫‪Français‬‬‫(قيم ضغط الدم مُقاسة بفواصل زمنية‬‫مقدارها خمس دقائق خالل اليوم)‬‫‪140‬‬‫‪140‬‬‫‪Español‬‬‫[مثال لتنوعات ضغط الدم خالل اليوم]‬‫ضغط دم‬‫انقباضي‬‫‪160‬‬‫‪160‬‬‫‪English‬‬‫يختلف ضغط الدم وف ًقا لس ّنك ونوعك وغير ذلك من العوامل‪ .‬وقد يتأثر بنظامك البيولوجي خالل اليوم وكذلك‬‫بوضعية جلوسك‪ ،‬واألنشطة الفيزيائية‪ ،‬واألنشطة العقلية‪ ،‬ومستوى اإلجهاد‪ ،‬وحتى بدرجة حرارة الجو‪ .‬ويتنوع‬‫ضغط الدم لدى الشخص الصحيح بصورة عامة خالل اليوم‪.‬‬‫‪06_CHU304_2401_Ara.indd 14‬‬‫‪2‬‬‫إذا قمت بالضغط على مفتاح الذاكرة‬‫مرة أخرى‪ ،‬تظهر بيانات القياس السابقة‪.‬‬‫في كل مرة تقوم بالضغط على المفتاح‪ ،‬تظهر بيانات القياس بالترتيب من األحدث إلى األقدم‪.‬‬‫* نموذج للشاشة عند الضغط على مفتاح الذاكرة‬‫‪.‬‬‫‪English‬‬‫‪Español‬‬‫يظهر رقم البيانات قبل نتائج‬‫القياس‪.‬‬‫•‬‫‪Français‬‬‫القيمة المخزنة في الذاكرة مر ّقمة بترتيب القياسات‪ .‬على سبيل المثال‪ ،‬عند تخزين ‪ 99‬مجموعة من البيانات في‬‫الذاكرة‪ ،‬فإن البيانات رقم ‪ 1‬تمثل أحدث بيانات‪.‬‬‫‪99 ،98 ...‬‬‫رقم البيانات‪... ،3 ،2 ،1 :‬‬
‫‪Ç‬‬
‫‪Ç‬‬
‫أقدم بيانات‬‫أحدث بيانات‬‫كيفية حذف البيانات‬‫‪Pyсский‬‬‫‪ 1‬عندما يتم عرض أي نتيجة (باستثناء متوسط القراءة آلخر ثالث قياسات)‪ ،‬اضغط مع االستمرار على مفتاح‬‫ثوان‪.‬‬‫ألكثر من ‪3‬‬‫الذاكرة‬‫ٍ‬‫سيتم حذف كافة البيانات المخزنة في الذاكرة‪.‬‬‫اضغط مع االستمرار‬‫ثوان‬‫لمدة ‪3‬‬‫ٍ‬‫‪हिन्दी‬‬‫تم الحذف‬‫تظهر رسالة تخبرك بأنه قد تم حذف‬‫كافة البيانات من الذاكرة‪.‬‬‫حول "ضغط الدم"‬‫ما هو ضغط الدم؟‬‫(انبساطي)‬‫(انقباضي)‬‫القلب عبارة عن مضخة توزع الدم على‬‫الجسم بالكامل‪ .‬ويتم ضخ الدم من القلب‬‫األُذين األيسر‬‫األُذين األيسر‬‫بضغط مستمر إلى الشرايين‪ .‬هذا الضغط‬‫يُسمى ضغط الدم الشرياني ويمثل ضغط الدم‬‫تتم استعادة الحالة العادية البُطين‬‫البُطين‬‫األيسر‬‫لديك بصورة عامة‪ .‬ويُشار إلى ضغط الدم‬‫لجدار الشريان‪.‬‬‫يتمدد جدار الشريان‪ .‬األيسر‬‫بعدة أنواع من الضغط‪ ،‬بما في ذلك الضغط‬‫انبساط‬‫انقباض‬‫االنقباضي الذي يحدث عندما يضخ القلب الدم‪،‬‬‫والضغط االنبساطي الذي يحدث عندما يعود‬‫الدم إلى القلب‪.‬‬‫الشريان‬‫‪ -‬العربية ‪- 13‬‬‫‪6/3/2024 12:29:21 PM‬‬‫الشريان‬‫‪06_CHU304_2401_Ara.indd 13‬‬‫يختلف ضغط الدم لديك وف ًقا لوضعية جلوسك وتوقيت القياس وغير ذلك من العوامل‪ .‬وبشكل مثالي‪ ،‬ينبغي عليك‬‫قياس ضغط الدم لديك بنفس وضعية الجلوس وفي نفس التوقيت من كل يوم‪.‬‬‫‪Pyсский‬‬‫‪Français‬‬‫■ أخذ القياس أثناء الجلوس‬‫*اجلس في وضع مريح لك مع وضع قدمك باستقامة على األرض‪،‬‬‫وال تضع ساقيك بشكل عرضي‪.‬‬‫ضع ذراعك على طاولة أو سطح مشابه مع بسط ساعدك‪.‬‬‫من المهم أن يكون الرباط على مستوى واحد مع قلبك‪.‬‬‫افتح يدك برفق على الطاولة بحيث تكون راحة يدك موجهة ألعلى‬‫وأصابعك مرتخية‪.‬‬‫ال تحرّ ك جسمك وال تتحدث أثناء أخذ القياس‪.‬‬‫■ القياس في وضعية االستلقاء‬‫‪Español‬‬‫نصائح للحصول على قياسات دقيقة‬‫‪English‬‬‫وصف اإلنذار التقني‬‫سيُظهر جهاز القياس "‪ "HI‬أو "‪ "Lo‬كإنذار تقني على الشاشة بدون تأخير إذا كان ضغط الدم المُحدد (انقباضي أو‬‫انبساطي) خارج المدي المحدد في الجزء الخاص بالمواصفات‪ .‬في هذه الحالة يجب عليك استشارة الطبيب أو التحقق‬‫مما إذا خالف تشغيلك للجهاز لتعليمات التشغيل‪.‬‬‫وضع حالة اإلنذار التقني (خارج نطاق القياس) هو وضع مُسبق الضبط في المصنع وال يُمكن تعديله أو إلغاء تفعيله‪.‬‬‫وضع اإلنذار هذا تم تعيينه كأولوية مُنخفضة وف ًقا لمعيار ‪ .IEC 60601-1-8‬اإلنذار التقني هو إنذار مؤقت وال يحتاج‬‫ثوان‪.‬‬‫إلى إعادة ضبط‪ .‬ستختفي اإلشارة التي ظهرت على الشاشة تلقائيًا بعد ‪8‬‬‫ٍ‬‫استرخ لتجنب وضع ضغط على الرباط‪.‬‬‫*‬‫ِ‬‫استعادة البيانات المخزنة في الذاكرة‬‫يمكن تخزين ‪ 99‬نتيجة قياس في الذاكرة‪.‬‬‫ُتحسب القيمة المتوسطة تلقائيًا لمساعدتك في إدارة صحتك‬‫اليومية‪.‬‬‫مفتاح الذاكرة‬‫‪ 1‬اضغط على مفتاح الذاكرة‬‫مرة واحدة‪.‬‬‫‪.‬‬‫* نموذج للشاشة عند الضغط على مفتاح الذاكرة‬‫تعرض اآلن شاشة ‪ LCD‬عدد النتائج في المخزن الحالي ويشير الرمز إلى متوسط القيمة الذي يظهر في‬‫الجزء السفلي من الشاشة‪ .‬ثم يتم عرض متوسط القيمة استنادًا إلى القياسات الثالثة األخيرة‪.‬‬‫‪ -‬العربية ‪- 12‬‬‫‪6/3/2024 12:29:20 PM‬‬‫‪हिन्दी‬‬‫■ يمكنك استخدام ذراعك األيمن لقياس ضغط الدم لديك‪.‬‬‫ربما تختلف قيمة ضغط الدم بنسبة ‪ 10‬مم زئبق تقريبًا عند قياسه على الذراع األيمن بدالً من األيسر (أو العكس)‪ .‬قم‬‫بقياس ضغط الدم لديك على نفس الذراع من كل يوم‪.‬‬‫‪06_CHU304_2401_Ara.indd 12‬‬

‫‪5‬‬‫عرض نتائج القياس‪.‬‬‫• ما أن يكتمل القياس‪ ،‬يتم تفريغ هواء الرباط وتظهر‬‫نتائج القياس (قيم ضغط الدم االنبساطي‪/‬االنقباضي‪،‬‬‫والنبض)‪.‬‬‫* تظهر ضغط الدم االنبساطي‪/‬االنقباضي‪ ،‬والنبض‬‫‪English‬‬‫بالتناوب‪.‬‬‫* ويتم تخزينها تلقائيًا‪ ،‬بشرط عدم وجود خطأ في‬‫نتائج القياس‪.‬‬‫مؤشر تصنيف ارتفاع ضغط الدم‬‫‪Español‬‬‫‪6‬‬‫إنهاء القياس‪.‬‬‫• اضغط على مفتاح " بدء‪/‬إيقاف (‪ ")START/STOP‬إليقاف تشغيل جهاز المراقبة‪.‬‬‫• بعد أخذ قياسات ضغط الدم‪ ،‬سيؤدي الضغط على مفتاح " بدء‪/‬إيقاف (‪ ")START/STOP‬إلى إيقاف‬‫تشغيل جهاز المراقبة‪ .‬مع ذلك‪ ،‬إذا لم تقم بالضغط على المفتاح‪ ،‬فإن وظيفة اإليقاف التلقائي بالشاشة سوف تقوم‬‫بإيقاف تشغيل الشاشة تلقائيًا بعد دقيقة واحدة‪.‬‬‫‪Français‬‬‫مؤشر عدم انتظام ضربات القلب‬‫إذا تم اكتشاف ضربات قلب غير منتظمة أثناء القياس‪ ،‬فسيظهر مؤشر عدم انتظام ضربات القلب‬‫(‪ )IHB‬بعد اكتمال القياس‪.‬‬‫‪Pyсский‬‬‫توجد حالتين سوف يتم من خاللهم عرض إشارة ‪:IHB‬‬‫‪ )1‬معامل التباين (‪ )CV‬لمدة النبض > ‪%25‬‬‫‪ )2‬الفرق بين فترة النبض المتقاربة ≤ ‪ 0.14‬ث‪ ،‬ويستغرق عدد هذه النبضات أكثر من ‪ 53‬في المائة من‬‫إجمالي عدد النبضات‪.‬‬‫•	قد ال يكون القياس صحيحً ا إذا ترددت ضربات القلب بصورة كبيرة أثناء القياس‪ .‬إذا ظهر مؤشر عدم انتظام‬‫ضربات القلب (‪ ،)IHB‬فقم بقياس ضغط الدم لديك مرة أخرى‪.‬‬‫تنبيه‬‫إذا ظهر مؤشر عدم انتظام ضربات القلب (‪ )IHB‬باستمرار‪ ،‬فنوصيك باستشارة طبيب‬‫حول حالتك الصحية‪.‬‬‫‪हिन्दी‬‬‫مؤشر تصنيف ارتفاع ضغط الدم‬‫يتم عرض قيمة ضغط الدم التي تم قياسها وف ًقا لمعايير منظمة الصحة العالمية (‪.)WHO‬‬‫(‬‫انظر صفحة ‪)2‬‬‫انقباضي (مم زئبق)‬‫الدرجة ‪ 3‬ارتفاع ضغط الدم (حاد)‬‫الدرجة ‪ 2‬ارتفاع ضغط الدم (متوسط)‬‫الدرجة ‪ 1‬ارتفاع ضغط الدم (طفيف)‬‫عال‪-‬عادي‬‫ٍ‬‫عادي‬‫مثالي‬‫انبساطي (مم زئبق)‬‫‪6/3/2024 12:29:20 PM‬‬‫‪110‬‬‫‪100‬‬‫‪90‬‬‫‪85‬‬‫‪80‬‬‫‪ -‬العربية ‪- 11‬‬‫‪180‬‬‫‪160‬‬‫‪140‬‬‫‪130‬‬‫‪120‬‬‫‪06_CHU304_2401_Ara.indd 11‬‬‫سطح مشاب ٍه بحيث يكون ارتفاع‬‫ضع ذراعك على طاولة أو‬‫ٍ‬‫مركز ذراعك مساويًا لمستوى قلبك‪.‬‬‫‪Español‬‬‫‪6‬‬‫اجعل ذراعك مرتخيًا وافتح يدك برفق‪ ،‬مع مراعاة أن تكون راحة اليد ألعلى‪.‬‬‫‪3‬‬‫ضع الرباط على نفس مستوى ارتفاع القلب‪.‬‬‫الضغط على مفتاح "‬‫بدء‪/‬إيقاف (‪.")START/STOP‬‬‫• يتم سماع صوت تنبيه ويتم عرض كل األرقام في الشاشة األولية‪ ،‬والتي توضح أن الشاشة تعمل بشكل‬‫طبيعي‪.‬‬‫• يتم ضغط الرباط تلقائيًا‪.‬‬‫* إذا كان الضغط بالوحدة غير كافٍ‪ ،‬فستقوم بإعادة الضغط تلقائيًا‪.‬‬‫* تقوم الوحدة تلقائيًا بضبط الضغط تبعًا للقياس السابق‪.‬‬‫‪हिन्दी‬‬‫‪4‬‬‫‪Français‬‬‫* ضع الرباط على ذراعك بعناية‪،‬‬‫مع مراعاة استخدام أسلوب التثبيت‬‫الصحيح‪ .‬ويتحقق أسلوب التثبيت‬‫الصحيح إذا تمكنت من إدخال إصبع‬‫واحد بين الرباط وذراعك بسهولة‪.‬‬‫‪Pyсский‬‬‫‪5‬‬‫‪English‬‬‫‪4‬‬‫اسحب طرف الرباط للخارج بحيث يكون الرباط ملفو ًفا‬‫بشكل مريح حول ذراعك ثم قم بإحكام قفل الطرف بالشريط‬‫الالصق‪.‬‬‫*	 يظهر عند اكتشاف نبض‪.‬‬‫*	عندما ترغب في إيقاف القياس‪ ،‬اضغط على مفتاح "‬‫يتم تفريغ هواء الرباط وإيقاف القياس‪.‬‬‫بدء‪/‬إيقاف (‪ .")START/STOP‬عندها‬‫* إذا كان للمستخدم السابق إعداد عامل ضغط عالي‪ ،‬فإن هذا الضغط العالي سوف يُستخدم للقياس الحالي‪.‬‬‫* إذا تجاوزت قيمة الضغط ‪ 260‬مم زئبق‪ ،‬أو شعرت أن الضغط غير طبيعي‪ ،‬أو كنت تريد إيقاف‬‫عملية القياس‪ ،‬فاضغط على مفتاح " بدء‪/‬إيقاف (‪ ")START/STOP‬مرة أخرى‪ .‬يتم تفريغ هواء‬‫الرباط وإيقاف تشغيل الجهاز‪.‬‬‫‪ -‬العربية ‪- 10‬‬‫‪6/3/2024 12:29:20 PM‬‬‫‪06_CHU304_2401_Ara.indd 10‬‬‫قياس ضغط الدم لديك‬‫‪1‬‬‫قم بتوصيل موصل خرطوم الهواء في‬‫الوحدة الرئيسية‪.‬‬‫‪English‬‬‫• ال تلو خرطوم الهواء أثناء القياس‪ ،‬حيث قد يُسبب خطأ باالنتفاخ‬‫أو إصابة وخيمة نتيجة لضغط الكفة المستمر‪.‬‬‫‪2‬‬‫فتحة خرطوم‬‫الهواء‬‫موصل خرطوم‬‫الهواء‬‫توصيل الرباط‬‫‪Español‬‬‫تأكد من اختيار حجم الرباط المناسب‪ ،‬قبل قيامك بوضع رباط قياس ضغط الدم الخاص بك‪:‬‬‫الرباط (الطراز‪ )SCN-008 :‬مناسب للذراع الذي محيطه‪ 22 :‬سم – ‪ 30‬سم‪.‬‬‫طوق النطاق العريض (الطراز‪ (SCW-009 :‬يالئم محيط الذراع‪ 22 :‬سم –‪ 24‬سم‪.‬‬‫‪Français‬‬

‫‪1‬‬
‫‪2‬‬

‫قم بفك الرباط وإدخال طرفه في الطوق المعدني بحيث يكون الجانب‬‫المزود بالشريط الالصق نحو الخارج‪.‬‬‫ضع الطوق حول زراعك بحيث يصل خرطوم الهواء إلى راحة اليد‪.‬‬‫‪Pyсский‬‬‫طوق معدني‬‫‪हिन्दी‬‬‫‪ :‬وضع الشريان‬‫‪3‬‬‫اضبط موضع الطوق لذلك يتم وضع خرطوم الهواء في وسط الذراع وتكون حافة الطوق على ارتفاع ‪ 1‬سم –‬‫‪ 2‬سم فوق الكوع‪.‬‬‫عار أو فوق مالبس‬‫* يجب وضع الرباط على ذراع ٍ‬‫داخلية خفيفة الوزن‪.‬‬‫* إذا كنت ترتدي فانلة ثقيلة‪ ،‬فيرجى نزعها‪.‬‬‫‪ 1‬سم إلى ‪ 2‬سم‬‫‪ -‬العربية ‪- 9‬‬‫‪6/3/2024 12:29:19 PM‬‬‫‪06_CHU304_2401_Ara.indd 9‬‬‫ضبط الوقت‬‫يم ّكنك إعداد الساعة من تسجيل القياسات بدقة‪.‬‬

‫‪5‬‬
‫‪6‬‬
‫*‬

‫‪ "MEMORY‬لضبط "الساعة"‪.‬‬‫اضغط على المفتاح "‬‫‪ "MEMORY‬لزيادة العدد بنسبة واحد‪ .‬اضغط على‬‫اضغط على المفتاح "‬‫المفتاح " ‪ "START/STOP‬لتأكيد الضبط‪ .‬يتم ضبط مؤشر "الساعة" ويومض‬‫مؤشر "الدقائق"‪.‬‬‫‪ "MEMORY‬لضبط "الدقائق"‪.‬‬‫اضغط على المفتاح "‬‫‪ "MEMORY‬لزيادة العدد بنسبة واحد‪ .‬اضغط على‬‫اضغط على المفتاح "‬‫المفتاح " ‪ "START/STOP‬لتأكيد الضبط‪.‬‬‫اضغط على " ‪ "START/STOP‬مع إشارة ضبط الوقت على الراديو‪ ،‬وغير‬‫ذلك إلكمال ضبط الساعة‪.‬‬‫‪ -‬العربية ‪- 8‬‬‫‪6/3/2024 12:29:19 PM‬‬‫‪हिन्दी‬‬‫‪4‬‬‫‪ "MEMORY‬لضبط "اليوم"‪.‬‬‫اضغط على المفتاح "‬‫‪ "MEMORY‬لزيادة العدد بنسبة واحد‪ .‬اضغط على‬‫اضغط على المفتاح "‬‫المفتاح " ‪ "START/STOP‬لتأكيد الضبط‪ .‬يومض مؤشر "اليوم" و"الساعة" ‪.‬‬‫‪Pyсский‬‬‫‪3‬‬‫‪ "MEMORY‬لضبط "الشهر"‪.‬‬‫اضغط على المفتاح "‬‫‪ "MEMORY‬لزيادة العدد بنسبة واحد‪ .‬اضغط على المفتاح‬‫اضغط على المفتاح "‬‫" ‪ "START/STOP‬لتأكيد الضبط‪ .‬يتم ضبط "الشهر" و يومض مؤشر "اليوم"‪.‬‬‫* يمكنك إعادة التوجيه السريع لألعداد في الشاشة بواسطة الضغط مع االستمرار على‬‫مفتاح‬‫‪.MEMORY‬‬‫‪Français‬‬‫‪2‬‬‫اضغط مع االستمرار على المفتاحين " ‪ ”START/STOP‬و“‬‫‪ "MEMORY‬لثانيتين تقريبًا‪ ،‬وسوف تسمع صوت تنبيه ويومض مؤشر‬‫"الشهر" أوالً‪.‬‬‫‪Español‬‬‫‪1‬‬‫عندما تقوم بتركيب البطاريات للمرة األولى‪ ،‬سوف تعرض الشاشة الوقت‬‫والتاريخ على التوالي‪.‬‬‫‪English‬‬‫يمكن لجهاز المراقبة تسجيل وقت وتاريخ القياس باإلضافة إلى نتائج القياس‪ .‬ولذا‪ ،‬تأكد من ضبط الساعة بعد إدخال‬‫البطاريات‪ .‬اضبط التاريخ أوالً ثم اضبط الوقت‪.‬‬‫‪06_CHU304_2401_Ara.indd 8‬‬‫تحميل البطاريات‬‫(يجب تحميل البطاريات المرفقة مع الوحدة في حجيرتها قبل استخدام جهاز مراقبة ضغط الدم)‪.‬‬‫‪English‬‬‫‪1‬‬‫افتح غطاء حجيرة البطاريات‪.‬‬‫اضغط على الجزء العلوي لغطاء البطارية‪ ،‬واسحب الغطاء تجاهك للفتح‪.‬‬‫حجيرة البطارية‬‫‪Español‬‬‫غطاء البطارية‬‫‪Français‬‬‫* إذا ُترك جهاز مراقبة ضغط الدم غير مستخدم لفترة طويلة‪ ،‬فتجب إزالة البطاريات من‬‫الحجيرة‪.‬‬‫‪2‬‬‫‪Pyсский‬‬‫‪3‬‬‫ضع البطاريات في الحجيرة‪.‬‬‫ضع البطاريات مع األخذ في االعتبار رمزي القطبين‬‫الموجب والسالب و ‪.‬‬‫‪ 4‬بطاريات "‪"AA‬‬‫أغلق غطاء البطارية‪.‬‬‫‪हिन्दी‬‬‫إزالة البطاريات‬
‫•‬
‫•‬
‫•‬
‫•‬
‫•‬
‫•‬
‫•‬

‫أو ال يظهر شيء على الشاشة‪ ،‬استبدل البطاريات القديمة بأخرى جديدة‪ .‬يجب تغيير‬‫عندما تظهر األيقونة‬‫األربع بطاريات معًا في الحال‪.‬‬‫اضبط الساعة بعد شحن البطاريات‪.‬‬‫ال يتم حذف البيانات المخزنة في الذاكرة بتغيير البطاريات‪.‬‬‫ال تستخدم بطاريات قابلة إلعادة الشحن‪.‬‬‫ال تستخدم بطاريات قلوية (‪ )LR6‬وبطاريات منجنيز (‪ )R6P‬معًا‪.‬‬‫البطاريات المضمنة ألغراض توضيحية فقط‪ .‬قد يكون العمر االفتراضي للبطارية أقل من المحدد‪.‬‬‫عند التخلص من البطاريات‪ ،‬التزم باللوائح الحكومية أو قواعد المؤسسات العامة البيئية السارية في بلدك‪/‬منطقتك‪.‬‬‫‪ -‬العربية ‪- 7‬‬‫‪6/3/2024 12:29:19 PM‬‬‫‪06_CHU304_2401_Ara.indd 7‬‬‫تعريف األجزاء‬‫أسماء المكونات‬‫<الوحدة الرئيسية>‬‫‪English‬‬‫<الرباط>‬‫خرطوم الهواء‬‫الشاشة‬‫موصل خرطوم‬‫الهواء‬‫مفتاح "بدء‪/‬إيقاف‬‫(‪")START/STOP‬‬‫انتظار‬‫طوق معدني‬‫‪Pyсский‬‬‫فتحة خرطوم الهواء‬‫* انظر صفحة ‪ 7‬للتعرف على حجيرة البطارية‬‫<الشاشة>‬‫أيقونة البطارية‬‫‪हिन्दी‬‬‫جاهز لمؤشر النفخ‬‫شاشة ضغط الدم‬‫االنقباضي‬‫مؤشر تصنيف‬‫ارتفاع ضغط الدم‬‫أيقونة الذاكرة‬‫ضغط الدم االنبساطي‪/‬‬‫عرض النبض‬‫عرض التاريخ‪/‬الوقت‬‫* راجع صفحة ‪ 11‬لمؤشر ضربات القلب غير المنتظمة (‪.)IHB‬‬‫‪ -‬العربية ‪- 6‬‬‫‪6/3/2024 12:29:19 PM‬‬‫‪Français‬‬‫‪Español‬‬‫مفتاح "الذاكرة‬‫(‪")M‬‬‫مؤشر عدم انتظام ضربات‬‫القلب (‪*)IHB‬‬‫‪06_CHU304_2401_Ara.indd 6‬‬‫معلومات حول التخلص في الدول األخرى خارج االتحاد األوروبي‪.‬‬‫إذا كنت ترغب في التخلص من البطاريات المستعملة‪ ،‬فيرجى االتصال بالسلطات المحلية لديك أو الموزع الخاص بك‬‫والسؤال عن الطريقة الصحيحة للتخلص منها‪.‬‬‫مالحظة حول رمز البطارية‪.‬‬‫‪English‬‬‫قد يُستخدم الرمز باالشتراك مع رمز كيميائي‪ .‬وفي هذه الحالة‪ ،‬فهو متوافق مع المتطلبات المحددة بواسطة التوجيهات‬‫الخاصة بالمادة الكيميائية الموجودة‪.‬‬‫المعايير المرجعية األوروبية‪:‬‬‫يتطابق جهاز قياس ضغط الدم مع األنظمة ‪.EN1060-4 ،EN1060-3 ،EN1060-1 ،EN60601-1‬‬‫المعايرة‪:‬‬‫‪Español‬‬‫يوصى‪ ،‬بصفة عامة‪ ،‬بفحص جهاز المراقبة الخاص بقياس ضغط الدم كل عامين لضمان التشغيل الصحيح والدقة‬‫السليمة‪ .‬يرجى االتصال بالموزع‪.‬‬‫‪Français‬‬‫االسم‪:‬‬‫العنوان‪:‬‬‫المصنع‬‫االسم‪:‬‬‫العنوان‪:‬‬‫‪CITIZEN SYSTEMS JAPAN CO., LTD.‬‬‫‪6-1-12, Tanashi-cho, Nishi-Tokyo-shi,‬‬‫‪Tokyo 188-8511, Japan‬‬‫‪ANDON HEALTH CO., LTD.‬‬‫‪No.3 Jinping Street, YaAn Road, Nankai District, Tianjin 300190, China‬‬‫‪Pyсский‬‬‫المندوب باالتحاد األوروبي‪:‬‬‫‪EMERGO EUROPE‬‬‫االسم‪:‬‬‫العنوان ‪Westervoortsedijk 60, 6827 AT Arnhem, The Netherlands :‬‬‫• للحصول على البيانات الفنية‪ ،‬يرجى االتصال بمندوبنا باالتحاد األوروبي‬‫‪हिन्दी‬‬‫صُنع في الصين‬‫‪ -‬العربية ‪- 5‬‬‫‪6/3/2024 12:29:18 PM‬‬‫‪06_CHU304_2401_Ara.indd 5‬‬‫‪ .7‬ال تقم أبدًا بتنظيف جهاز مراقبة ضغط الدم باستخدام كحول أو تنر أو بنزين‪ ،‬فقد يؤدي هذا إلى تلف الجهاز‪.‬‬‫‪ .8‬لتنظيف الرباط‪ ،‬امسحه بقطعة قماش مبللة‪ .‬تجنب الحك الخشن‪ ،‬فسوف يؤدي هذا إلى تسرب الهواء‪ .‬واحذر‬‫كذلك من دخول الماء إلى خرطوم الهواء‪.‬‬‫‪ .9‬يوصى بتنظيف الكفة بعد كل ‪ 200‬مرة من االستخدام‪.‬‬‫تحذير‬‫‪हिन्दी‬‬‫طبيعة الصيانة وتكرارها‪:‬‬‫هذا المنتج مُصمم لالستخدام لفترة طويلة من الوقت؛‬‫ورغم ذلك‪ ،‬يوصى عمومًا بفحصه كل عامين لضمان التشغيل واألداء بطريقة صحيحة‪.‬‬‫حماية البيئة الطبيعية‪:‬‬‫يُرجى المساعدة في حماية البيئة الطبيعية باتباع لوائح إعادة التدوير الوطنية و‪/‬أو المحلية عند التخلص من البطارية‬‫والمنتج في نهاية عمر استخدامه‪.‬‬‫عالمة ‪WEEE‬‬‫إذا أردت التخلص من هذا المنتج‪ ،‬فال تخلطه بالفضالت المنزلية العامة‪ .‬إذ توجد أنظمة تجميع‬‫منفصلة للمنتجات اإللكترونية المستعملة وف ًقا لقوانين بموجب توجيهات نفايات المعدات الكهربائية‬‫واإللكترونية (التوجيهات ‪ )2012/19/EU‬وهي سارية داخل االتحاد األوروبي فقط‪.‬‬‫‪ -‬العربية ‪- 4‬‬‫‪6/3/2024 12:29:18 PM‬‬‫‪Pyсский‬‬‫أُخضع الجهاز لالختبار وحصل على التوافقية وف ًقا لمعيار ‪ EN60601-1-2‬فيما يتعلق‬‫باالضطرابات الكهرومغناطيسية (‪ .)EMD‬وال يضمن هذا‪ ،‬بأي شكل من األشكال‪ ،‬عدم‬‫تأثر الجهاز بالتداخل الكهرومغناطيسي‪ .‬ولذا‪ ،‬تجنب استخدام الجهاز في البيئات ذات‬‫الكهرومغناطيسية المرتفعة‪.‬‬‫‪Français‬‬‫شرح الرموز‬‫‪Español‬‬‫ال يسمح بتعديل هذه المعدات‪.‬‬‫‪:‬تهدف العالمة ‪ CE‬إلى توفير المعلومات لمفتشي السوق في البلدان األعضاء في االتحاد‬‫األوروبي‪.‬‬‫‪ :‬جهاز من النوع ‪( BF‬الكفة من النوع ‪ BF‬ألجزاء المعدات الطبية‪).‬‬‫‪ :‬أبقه جا ًفا‬‫‪ :‬تحذير‬‫‪ :‬ارجع إلى دليل التعليمات قبل االستخدام (لون خلفية اإلعالن ‪:‬أزرق‪ .‬الرمز الرسومي لإلعالن‪ :‬أبيض‪).‬‬‫‪ :‬توجيهات نفايات المعدات الكهربائية واإللكترونية المتعلقة بامتثال األجهزة‬‫‪ :‬جهة التصنيع‬‫‪ :‬رمز لـ "التفويض األوروبي"‬‫تحذير‬‫‪English‬‬‫‪ .4‬ال تحاول فك الجهاز أو الرباط‪ .‬اسأل مركز الخدمة المُحدد من قبل جهة التصنيع‪.‬‬‫‪ .5‬ال تقم بثني الرباط أو خرطوم الهواء بصورة زائدة عن الحد‪.‬‬‫‪ .6‬إذا كان جهاز القياس والكفة شديدي االتساخ‪ّ ،‬‬‫نظفهما بمسحهما بقطعة قماش مُبللة بكحول مُعقم أو مُطهر م ُتعادل‪.‬‬‫ُثم امسحهما بقطعة قماش جافة‪ .‬عند الحاجة بالمستشفى‪ ،‬يجب تعقيم الكفة مرتين أسبوعيًا‪.‬‬‫‪06_CHU304_2401_Ara.indd 4‬‬‫احتياطات االستخدام والصيانة‬‫احتياطات االستخدام‬‫‪English‬‬‫‪Español‬‬‫‪Français‬‬‫‪Pyсский‬‬‫‪हिन्दी‬‬‫‪ .1‬إذا كنت تعاني من مرض بالقلب أو ارتفاع ضغط الدم أو مرض آخر متعلق بالدورة الدموية‪ ،‬فاستشر طبيبك قبل‬‫استخدام جهاز المراقبة‪.‬‬‫‪ .2‬قد يؤدي النفخ بضغط مرتفع إلى إحداث كدمة في المكان المُستخدم فيه الرباط‪.‬‬‫فإذا شعرت أن ضغط الرباط غير طبيعي أو واجهت أي أمر غير عادي أثناء استخدام الرباط‪ ،‬فقلل الضغط فورً ا‬‫بالضغط على مفتاح "بدء‪/‬إيقاف (‪ ،")START/STOP‬أو فصل الرباط‪ ،‬أو نزع موصل خرطوم الهواء من جهاز‬‫المراقبة ثم استشارة منفذ البيع الذي اشتريت منه الجهاز‪.‬‬‫‪ .3‬إذا رأيت أن القياس غير طبيعي أو يجعلك تشعر بأنك مريض‪ ،‬فتوقف عن استخدام الجهاز واستشر طبيبك‪.‬‬‫‪ .4‬قد ال يمكن قياس ضغط الدم ألي شخص يعاني من ضعف في النبض أو عدم انتظام ضربات القلب‪.‬‬‫‪ .5‬قد يؤدي القياس المتكرر لضغط الدم والنفخ الزائد عن الحد لفترة طويلة إلى مشكالت مثل االحتقان أو التورم‬‫عند بعض األشخاص‪.‬‬‫‪ .6‬لن تعطي القياسات المتكررة بصورة مستمرة لضغط الدم نتائج دقيقة‪ .‬ولذا‪ ،‬عليك أن تسمح بفاصل زمني مدته‬‫حوالي دقيقة بين كل عملية قياس‪.‬‬‫‪ .7‬إذا كنت تعاني من مشكلة حادة في الدورة الدموية في ذراعيك‪ ،‬فاستشر طبيبك قبل استخدام الجهاز‪ .‬فقد يكون‬‫عدم القيام بذلك خطرً ا على صحتك‪.‬‬‫‪ .8‬قد ال يمكن القياس ألي شخص يعاني من ضعف ضخ الدم إلى المنطقة التي سيتم فيها القياس أو شخص يعاني‬‫من ضربات قلب غير منتظمة بصورة دائمة‪ .‬استشر طبيبك واطلب نصيحته حول استخدام الجهاز من عدمه‪.‬‬‫‪ .9‬ال تقم بلف الرباط حول ذراع مجروح‪.‬‬‫‪ .10‬ال تقم بلف الرباط حول معصم به محقن أوردة‪ ،‬أو ذراع مستخدَم في نقل الدم كجزء من عالج طبي‪ .‬فيمكن أن‬‫يؤدي القيام بذلك إلى اإلصابة أو إلى حادث خطير‪.‬‬‫‪ .11‬ال تقم بمشاركة الكفة مع أشخاص أخرين مصابين بالعدوى لتجنب انتقال العدوى‪.‬‬‫‪ .12‬إذا كان لديك أي شك حول استخدام الكفة على الذراع المجاور لمنطقة عملية استئصال الثدي‪ ،‬استشر طبيبك‪.‬‬‫‪ .13‬ال تستخدم الجهاز بالقرب من الغازات القابلة لالشتعال مثل الغازات المُستخدمة في التخدير‪ .‬فيمكن أن يؤدي‬‫القيام بذلك إلى إشعال الغازات والتسبب في وقوع انفجار‪.‬‬‫‪ .14‬ال تستخدم الجهاز في بيئات غنية باألكسجين مثل غرفة الضغط العالي أو خيمة األكسجين بالمستشفيات‪ .‬فيمكن‬‫أن يؤدي القيام بذلك إلى إشعال األكسجين والتسبب في نشوب حريق‪.‬‬‫‪ .15‬ال تستخدم األجهزة المحمولة بالقرب من الجهاز‪ ،‬فقد يؤدي هذا إلى تعطل الجهاز‪.‬‬‫‪ .16‬إذا كنت تستخدم مُنظمًا قلبيًا‪ ،‬فاستشر طبيبك قبل استخدام جهاز المراقبة‪.‬‬‫‪ .17‬تأكد من استخدام هذه الوحدة لقياس ضغط الدم فقط‪ .‬وال تستخدمها ألي غرض آخر‪.‬‬‫‪ .18‬تأكد من استخدام األجزاء والملحقات المرخصة فقط‪ .‬فقد تؤدي األجزاء والملحقات غير المرخص باستخدامها‬‫مع الجهاز إلى تلف الوحدة‪.‬‬‫‪ .19‬ال تستخدم هذه الوحدة على الرضع‪.‬‬‫‪ .20‬قد ال يمكن قياس ضغط الدم ألي شخص يعاني من حاالت شائعة لضربات القلب غير المنتظمة مثل الضربات‬‫األُذينية أو البُطينية السابقة ألوانها أو الرجفان األُذيني‪.‬‬‫احتياطات الصيانة‬‫‪ .1‬ال تقم بتخزين جهاز مراقبة ضغط الدم في األماكن المعرضة لضوء الشمس المباشر أو درجات الحرارة العالية‬‫(أعلى من ‪ °55‬درجة مئوية)‪ ،‬أو درجات الحرارة المنخفضة (أقل من ‪ °20-‬درجة مئوية)‪ ،‬أو الرطوبة النسبية‬‫العالية (أعلى من ‪ )%90‬أو كميات كبيرة من األتربة‪.‬‬‫تأكد من تخزين جهاز مراقبة ضغط الدم بعيدًا عن متناول األطفال وبعيدًا عن الحيوانات األليفة و‪ /‬أو اآلفات‪.‬‬‫‪ .2‬ال ُتسقط جهاز مراقبة ضغط الدم أو الرباط أو ُتعرّ ضه الهتزازات أو صدمات أخرى‪.‬‬‫‪ .3‬قم بإزالة البطاريات إذا تركت الجهاز غير مُستخدم لفترة طويلة‪.‬‬‫‪ -‬العربية ‪- 3‬‬‫‪6/3/2024 12:29:18 PM‬‬‫‪06_CHU304_2401_Ara.indd 3‬‬‫الدرجة ‪ 3‬ارتفاع ضغط الدم (حاد)‬‫الدرجة ‪ 2‬ارتفاع ضغط الدم (متوسط)‬‫‪160‬‬‫الدرجة ‪ 1‬ارتفاع ضغط الدم (طفيف)‬‫‪130‬‬‫‪120‬‬‫عال‪-‬عادي‬‫ٍ‬‫عادي‬‫مثالي‬‫‪80‬‬‫‪90 85‬‬‫‪100‬‬‫‪ 110‬انبساطي (مم زئبق)‬‫تعريفات وتصنيفات مستويات ضغط الدم‬‫(مثالي‪/‬عادي‪/‬عال‪-‬عادي‪/‬الدرجة ‪ 1‬ارتفاع ضغط‬‫* تم تجهيز هذه الوحدة والتي توضح تصنيف ضغط الدم مرئيًا‬‫ٍ‬‫الدم‪/‬الدرجة ‪ 2‬ارتفاع ضغط الدم‪/‬الدرجة ‪ 3‬ارتفاع ضغط الدم) للنتيجة بعد كل قياس‪.‬‬‫‪ .11‬الغرض من االستخدام‬‫•هذا الجهاز هو غير باضع ومخصص لمراقبة ضغط الدم بواسطة طريقة قياس الذبذبات‪ .‬يمكن لهذا الجهاز‬‫قياس ضغط الدم االنقباضي (‪ ،)SYS‬وضغط الدم االنبساطي (‪ ،)DIA‬ومعدل النبض تلقائيًا‪.‬‬‫•تم تصميم هذا الجهاز الستخدامه لفحص الحالة الصحية الشخصية في المنزل تحت إشراف طبيب ولم يتم‬‫ً‬‫جهازا تشخيصيًا‪.‬‬‫تصميمه ليكون‬‫•لم يتم تصميم هذا الجهاز لالستخدام مع األطفال حديثي الوالدة أو الرضع‪ ،‬ولم يتم تصميمه لقياس الدوران‬‫التلقائي‪.‬‬‫قم بقياس ضغط الدم لديك في نفس التوقيت من كل يوم‪.‬‬‫*‬‫يتغير ضغط الدم لديك طوال الوقت‪ .‬ويعني هذا أن البيانات التي يتم تجميعها على مدى فترة طويلة لها أهمية‬‫أكبر من البيانات الناتجة من قياس واحد‪ .‬ولهذا‪ ،‬يجب عليك قياس ضغط الدم لديك بشكل يومي‪ .‬وبشكل مثالي‪،‬‬‫ينبغي عليك قياس ضغط الدم لديك في نفس التوقيت من كل يوم‪ ،‬حيثما أمكن‪.‬‬‫‪ -‬العربية ‪- 2‬‬‫‪6/3/2024 12:29:18 PM‬‬‫‪हिन्दी‬‬‫‪140‬‬‫مالحظات‪:‬‬‫الرسم ليس دقي ًقا‪ ،‬ولكن يمكن استخدامه‬‫كمرشد في فهم قياسات ضغط الدم غير‬‫الباضعة‪ .‬الجهاز مُصمم لالستخدام من قبل‬‫البالغين فقط‪.‬‬‫‪Pyсский‬‬‫‪Français‬‬‫‪180‬‬‫‪Español‬‬‫انقباضي (مم زئبق)‬‫‪English‬‬‫‪ .4‬ال تقم بقياس ضغط الدم لديك بعد االستحمام أو الشرب‪.‬‬‫‪ .5‬قم بقياس ضغط الدم لديك عندما تكون درجة حرارة الغرفة حوالي ‪ °20‬مئوية‪ .‬وال تقم بقياس ضغط الدم لديك‬‫عندما تكون درجة حرارة الغرفة باردة للغاية (أدنى من ‪ °10‬مئوية) أو ساخنة للغاية (أعلى من ‪ °40‬مئوية)‪.‬‬‫‪ .6‬ال تحاول قياس ضغط الدم لديك بعد شرب القهوة أو الشاي أو بعد التدخين مباشر ًة‪.‬‬‫أبق مركز الرباط في مستوى القلب وال تحرّ ك ذراعك‬‫‪ .7‬قم بقياس ضغط الدم لديك عندما تكون مسترخيًا وثاب ًتا‪ِ .‬‬‫وال تتحدث‪.‬‬‫‪ .8‬ال تقم بقياس ضغط الدم لديك عندما يكون الرباط ملفو ًفا على ذراعك منذ بضع دقائق أو أكثر‪ .‬فخالل ذلك‬‫الوقت‪ ،‬سيعمل ذراعك على رفع ضغط الدم ولن تحصل على قراءة صحيحة‪.‬‬‫‪ .9‬في حالة ظهور مؤشر ضربات القلب غير المنتظمة (‪ ،)IHB‬فقم باالسترخاء وقم بقياس ضغط الدم مرة أخرى‪.‬‬‫إذا ظهر مؤشر عدم انتظام ضربات القلب (‪ )IHB‬باستمرار‪ ،‬فنوصيك باستشارة طبيب حول حالتك الصحية‪.‬‬‫(‬‫انظر صفحة ‪)11‬‬‫‪ .10‬الشكل التالي يوضح تصنيفات ضغط الدم وف ًقا لمعايير منظمة الصحة العالمية‪.‬‬‫‪06_CHU304_2401_Ara.indd 2‬‬‫العربية‬‫المحتويات‬‫‪English‬‬‫‪Español‬‬‫‪Français‬‬‫‪ - 1‬مالحظات عامة‬‫‪ - 3‬احتياطات االستخدام والصيانة‬‫‪ - 6‬تعريف األجزاء‬‫‪ - 7‬تحميل البطاريات‬‫‪ - 7‬إزالة البطاريات‬‫‪ - 8‬ضبط الوقت‬‫‪ - 9‬قياس ضغط الدم لديك‬‫‪ - 12‬نصائح للحصول على قياسات دقيقة‬‫‪ - 12‬استعادة البيانات المخزنة في الذاكرة‬‫‪ - 13‬حول "ضغط الدم"‬‫‪ - 14‬س و ج حول ضغط الدم‬‫‪ - 15‬قبل طلب اإلصالح أو االختبار‬‫‪ - 16‬المواصفات‬‫‪Pyсский‬‬‫■ تأكد من أن لديك كافة المكونات التالية‪.‬‬‫•وحدة جهاز مراقبة ضغط الدم • دليل التعليمات‬‫• ‪ ٤‬بطاريات حجم ‪( AA‬خاصة بجهاز المراقبة)‬‫• الطوق (الطراز‪ )SCN-008 :‬لـ ‪CHU304‬‬‫• طوق النطاق العريض (الطراز‪ )SCW-009 :‬لـ ‪CHU305‬‬‫مالحظات عامة‬‫مالحظات عامة حول ضغط الدم وقياسه‬‫‪हिन्दी‬‬‫‪.1‬‬‫‪.2‬‬‫‪.3‬‬‫‪.4‬‬‫‪.5‬‬‫‪.6‬‬‫لتفريغ هواء الرباط بسرعة‪ ،‬اضغط على مفتاح "بدء‪/‬إيقاف (‪.")START/STOP‬‬‫تأكد من اختيار حجم الرباط المناسب‪ ،‬قبل قيامك بوضع رباط قياس ضغط الدم الخاص بك‪:‬‬‫الرباط (الطراز‪ )SCN-008 :‬مناسب للذراع الذي محيطه‪ 22 :‬سم – ‪ 30‬سم‬‫طوق النطاق العريض (الطراز‪ )SCW-009:‬يالئم محيط الذراع‪ 22 :‬سم – ‪ 42‬سم‪.‬‬‫عند ربط الطوق‪ ،‬قم بضبط موضع الطوق لذلك يتم وضع خرطوم الهواء في وسط الذراع وتكون حافة الطوق‬‫على ارتفاع ‪ 1‬سم – ‪ 2‬سم فوق الكوع‪.‬‬‫القياس الذاتي ليس عالجً ا! ولذا ينبغي عليك عدم تغيير الدواء الموصوف لك من الطبيب في أي حال من‬‫األحوال‪.‬‬‫كتحضير لقياس ضغط الدم لديك‪ ،‬يجب عليك أن تتبول ثم تظل ثاب ًتا نسبيًا لمدة ‪ 10‬إلى ‪ 15‬دقيقة قبل القياس‪.‬‬‫ال تضع تركيزك كله على نتائج قياس واحد‪ .‬بل استمر في تسجيل النتائج المتنوعة لضغط الدم لديك‪ .‬إذ يمكنك‬‫الحصول على صورة كاملة من خالل عدد كبير من القراءات‪.‬‬‫للحصول على قياسات ضغط دم صحيحة‬‫استرخ قبل قياس ضغط الدم لديك‪ .‬إذا كنت متوترً ا عند أخذ القياس‪ ،‬فلن‬‫‪ .1‬خذ خمسة أو ستة أنفاس عميقة ثم‬‫ِ‬‫تحصل على قراءة صحيحة‪.‬‬‫‪ .2‬سيرتفع ضغط الدم لديك إذا كنت قل ًقا أو منزعجً ا‪ ،‬أو كنت تعاني من قلة النوم أو اإلمساك‪ ،‬أو إذا كنت قد أجريت‬‫بعض التمارين أو تناولت وجبة‪.‬‬‫‪ .3‬قم بلف رباط المعصم حول ذراعك وإحكامه بحيث يمكن إدخال أصبع واحد بين الرباط والذراع‪.‬‬‫(‬‫انظر صفحة ‪)10 ،9‬‬‫‪ -‬العربية ‪- 1‬‬‫‪6/3/2024 12:29:18 PM‬‬‫‪06_CHU304_2401_Ara.indd 1‬‬

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

00_CHU304_305_2401.indb 4

8/3/2024 9:21:36 AM