GENERAL CATALOG

2024.4.B(E)Torque testers are available for a variety of tightening tools.
For managing electric screwdriver

We keep providing high performance andhigh quality products trusted by our customer.

Calibration certificates, Inspection certificates, Traceability system chart
For managing click type torque driver
All torque measuring instruments come with a calibration certificate, inspection certificate, andtraceability system chart free of charge.

Calibration service

Periodic calibration inspections are necessary to maintain the accuracy of torque measuring

For managing click type torque wrench

instruments. Since we are a specialized manufacturer that designs and manufactures torquemeasuring instruments as well as provides after-sales service, high calibration accuracy isguaranteed.

After service
For managing impact tool and torque wrench
If the torque measuring instrument malfunctioned, we will repair it. For discontinued products,
we will repair them if the date of discontinuation is within 5 years. However, there are cases inwhich repair may not be possible due to the production status of parts at our suppliers. Please
understand this in advance. We also offer periodic inspections of our measuring instruments.
Please contact the dealer where you purchased your instrument.

12Index (For managing the tightening tools)
Power source

0.002~8 N・m
0.01~10 N・m
0.002~8 N・m

〇

〇

〇

0.3~25 N・m
7~200 N・m〇〇〇

0.002~1500 N・m

〇

〇

〇

〇

〇

Page

WDI series

5

〇

〇

〇

DI-9M series

7

〇

〇

△

CD series

9

〇

〇

△

NDI series

11

〇

〇

DI-4B-25

13〇△

DWT-200

15

〇

〇

〇

〇

WDIS-IP series

17

WDISR-IP series

19

WDISR-IPS series

1.00~1500 N・m
0.3~500 N・m

3

Model

WDIS-IPS series

0.002~20 N・m

〇

1000

RS232C0.002~25 N・m

〇
〇

100

USB

〇

10

Dry cell battery〇

1

Internal battery

Torque wrench〇

0.1

AC adapter

Torque driver〇

Impact tool

Electric screwdriver〇

0.01

COM type

〇

〇
〇
〇

〇

〇
〇

〇

〇
〇

WIMS-IP series

21

〇

〇

〇

DI-1M-IP series

23

〇

〇

△

4

For managing the tightening tools (integrated type)
Torque Tester WDI series
Electric screwdriver

Torque driver

Product specification detailMeasurement

Torque wrench

Code

joint

OW

SJ

OW joint

SJ joint

Contents

Package

Code

A

C

Attache case

Cardboard caseContents

Destination

This series are available for various tightening tools.
WDI-10HR / 100HR allows selection of measurement joints from SJ joint or OW joint.

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

FeaturesWeight unit (kgf･cm)

unit

Ordering number■ Easy-to-read color LCD display■ Various communication commands are available■ High resolution mode (only for WDI-10HR/100HR) Model-

Measurement
joint

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

-

Pack
age

-

Destina

Plug

tion

type

1
A type2
A type(China)3

4

5

C typeF typeG type■ Equipped with RS232C and USB port
Model

WDI-10HR

WDI-100HR

WDI-250

Normal

0.020〜1.000 N･m

0.20〜10.00 N･m

0.20〜25.00 N・m

High resolution

2.0〜999.9 ｍN･m

0.020〜9.999 N･m－

Measurement Range

OW joint

Cube joint

AC adapter［p.42］［p.43］［p.41］±0.5% (If 499 digit or less, ±3 digit.)

Accuracy
PowerInternal battery (Ni-MH)

Dimensions

160(W) × 145(D) × 58(H)

Accessories

AC adapter 6A-061WP12

If order OW joint

Accessories (OW joint type)Measurement joint OW-025

Measurement joint OW-20

Measurement joint OW-10

Measurement joint OW-60

Cube joint SC-1

Cube joint SC-2

Accessories (SJ joint type)
SJ joint

HEX attachment［p.42］［p.43］［p.43］Bit (4-pack)

Cube joint

AC adapter［p.44］［p.43］［p.41］* Each measurement joint comes with 2 spare screws.

Measurement joint SJ-10K

Measurement joint SJ-200

Measurement joint SJ-10K

Measurement joint SJ-50

HEX attachment HS6-M3 (X3) HEX attachment HS6-M3 (X3) HEX attachment HS6-M4(X3)

If order SJ joint

HEX attachment HS6-M4 (X3)

Joint cap W-CAP-13

Grease JT-GREASE

Joint cap W-CAP-13 (X2)

Grease JT-GREASE

Bit(4-pack) H6BIT-SET

Grease JT-GREASE

Socket bit H17SCKTBIT-A20

Cube joint SC-1

Bit(4-pack) H6BIT-SET

Bit(4-pack) H6BIT-SET

Cube joint SC-2

Cube joint SC-3

5

GreaseSocket but［p.43］［p.45］* Attached to SJ joint

Measurement joint SJ-50

Joint cap W-CAP-13

Joint cap

6

Torque Tester DI-9M series
Electric screwdriver

Product specification detail

Torque driver

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

1

Plug

A typetype

Accessories

Come with OW joint that do not require rewinding

OW joint

Cube joint

AC adapter［p.42］［p.43］［p.41］

DI-9M-08 / 8 is a torque tester which is ideal for daily inspections of electric drivers and torque drivers.

Features
Ordering number■ Equipped with pass/fail function■ Displays average, maximum and minimum value

Model■ Selectable measurement units- Destination■ Memory size is 800 data
■ Equipped with USB port
Model
Measurement Range
Accuracy

DI-9M-08

DI-9M-8

2.0~800.0 mN･m

0.020~8.000 N･m±0.5% (If 499 digit or less, ±3 digit.)

PowerInternal battery (Ni-MH)

Dimensions

160(W) × 125(D) × 55(H)Accessories

Measurement joint OW-025

Measurement joint OW-20

Measurement joint OW-10

Measurement joint OW-60

Cube joint SC-2
AC adapter 6A-061WP12* Each measurement joint comes with 2 spare screws.

7

8

2
A type(China)3

4

5

C typeF typeG typeTorque Tester CD series

Product specification detailElectric screwdriver

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

1

Plug

A typetype

2
A type(China)3

4

5

C typeF typeG typeAccessories

Come with SJ joints for conventional measurement methods

SJ joint

HEX attachment

CD-10M / 100M is a torque tester which is ideal for daily inspections of electric drivers.

Ordering number■ Equipped with pass/fail function■ Displays average, maximum and minimum value

Model■ Selectable measurement units［p.43］［p.43］

AC adapter- Destination■ Memory size is 800 data

［p.41］■ Equipped with USB port
Model
Measurement Range
Accuracy

CD-10M

CD-100M

0.010~1.000 N･m

0.10~10.00 N･m±0.5% (If 199 digit or less, ±2 digit.)

PowerInternal battery (Ni-MH)

Dimensions

160(W) × 125(D) × 55(H)Measurement joint SJ-50
Measurement joint SJ-10

Measurement joint SJ-10K

HEX attachment HS5-M3 (X3) HEX attachment HS6-M3 (X3)

Accessories

Joint cap W-CAP-13

HEX attachment HS6-M4 (X3)

Grease JT-GREASE

Joint cap W-CAP-13 (X2)

Bit(4-pack) H5BIT-SET

Grease JT-GREASE
Bit(4-pack) H6BIT-SET

AC adapter 6A-061WP12

9GreaseBit (4-pack)［p.43］［p.44］* Attached to SJ joint［p.42］

Features

Joint cap

10

Torque Tester NDI series
Electric screwdriver

Product specification detail

Torque driver

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

1

Plug

A typetype

2
A type(China)3

4

5

C typeF typeG typeAccessories

Integrate a clock function ideal for traceability

OW joint

Cube joint

USB cable

AC adapter［p.42］［p.43］［p.49］［p.41］

NDI-80CN / 800CN is a torque tester which is ideal for daily inspections of electric drivers and torque drivers.

Features
Ordering number■ Easy-to-read color LCD display■ Record “measurement date” andModel“measurement time” together- OW - A - Destination■ Advanced external control functions(Published the communication I/F as well)■ Powered by AC adapter (or dry cell battery)■ Equipped with RS232C and USB port
Model
Measurement Range

NDI-80CN

NDI-800CN

2.0~800.0 mN･m

0.020~8.000 N･m

Accuracy±0.5% (If 499 digit or less, ±3 digit.)

Power

AC adapter (or dry cell battery)Dimensions

Accessories

160(W) × 145(D) × 58(H)Measurement joint OW-025

Measurement joint OW-20

Measurement joint OW-10

Measurement joint OW-60

Cube joint SC-1

Cube joint SC-2

USB CABLE USB-ECOM510
AC adapter 6A-061WP12

Dry cell battery (4 pieces)* Each measurement joint comes with 2 spare screws.

11

12

Dry cell battery

Torque Tester DI-4B-25
Electric impact tool

Oil pulse tool

Product specification detailElectric screwdriver

Air impact tool

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

1

Plug

A typetype

2
A type(China)3

4

5

C typeF typeG typeAccessories

Manage the blow force and number that affect tightening

HEX14 joint

GreaseSocket bit

HEX socket

AC adapter［p.42］［p.43］［p.45］［p.45］［p.41］

DI-4B-25 is a torque tester ideal for daily inspections of impact tool and electric driver.

Features
Ordering number■ Easy-to-see high brightness LED■ Simultaneously measure the blow force andModelblow number- Destination■ Come with the soft and hard joint■ Equipped with USB port
Model

DI-4B-25

Measurement range
Except impact tool
Impact tool
Accuracy

0.3~25.00 N･m
0.3~15.00 N･m (Note)±0.5% (If 499 digit or less, ±3 digit.)

PowerInternal battery (Ni-MH)

Dimensions

150(W) × 220(D) × 74(H)Measurement joint HEX14-H250
Measurement joint HEX14-S250

Accessories

Grease JT-GREASE
Socket bit H17SCKTBIT-A20
HEX socket H17SOCKET-SQ9.5
AC adapter 6A-061WP12(Note) When using an impact tool, please make sure that the tightening torque value of used tool does notexceed the measurement range of the tester.

13

14

Torque Tester DWT-200

Product specification detail

Torque wrench

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)measurement
unit

Yard-pound unit (lbf･in)Accessories
Conversion adapter

Simple model for torque wrench management

Dry cell battery［p.45］

DWT-200 is a torque tester which is ideal for daily inspections of torque wrench.

Features
Ordering number■ Easy-to-see high brightness LED■ Ideal for work training
PP ModePD ModeModel

Measure the maximum torque- Destination

Measure the torquewhen feel a click■ Powered by dry cell battery
Model
Measurement Range
AccuracyPower
Dimensions
Accessories

DWT-200
7.0~200.0 N･m±0.5% (If 499 digit or less, ±3 digit.)
Dry cell battery
200(W) × 120(D) × 96(H)Conversion adapter BAE-34

Dry cell battery (4 pieces)

15

SI unit (N･m)
Weight unit (kgf･cm)

16

For managing the tightening tools (separated type)
Torque Tester WDIS/WDISR-IP series
Torque wrench

Nut runner

Product specification detail

Detector cable

Code

Torque driver

L1

L2

L4

0.9m（10P-1MF）

1.9m（10P-2MF）

3.9m（10P-4MF）

Contents

Package

Code

A

C

Attache case

Cardboard caseContents* For IP200 / 500 / 1500, only cardboard is available

Prepare the “wide measurement range”, “5 kinds of socket size”

Jx

Ox

Enable to choose the indicator to suit the usage scene

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)Power source can be selected from either AC adapter (WDIS) or internal battery (WDISR)

measurement

Destination

Code

SI unit (N･m)
Weight unit (kgf･cm)

unit

Features

Code■ Compact type not required a lot of space■ Various communication commands are available

Model

-

Detector■ Choice of power source and detector calble length

cable

age

tion

WDIS-IP05

WDIS-IP5

WDIS-IP50

WDIS-IP200

WDIS-IP500

WDIS-IP1500

Measurement

2.0〜500.0

0.020〜5.000

0.20〜50.00

2.0〜200.0

2.0〜500.0

20〜1500

ｍN･m

N･m

N･m

N･m

N･m

N･m

Accuracy±0.5% (If 499 digit or less, ±3 digit.)

Power

AC adapter (or dry cell battery) * Tester power will be interlocked with external power source

Model

WDISR-IP05

WDISR-IP5

WDISR-IP50

WDISR-IP200

Measurement

2.0〜500.0

0.020〜5.000

0.20〜50.00

2.0〜200.0

2.0〜500.0

20〜1500

ｍN･m

N･m

N･m

N･m

N･m

N･m

Power

Internal battery (Ni-MH)

Accessories

Rubber footHandle［p.47］［p.49］［p.49］

Indicator : 103(W) X 108(D) X 38(H) , Detector : See to p.57
Detector cable

HandleRubber foot B8-4

CPH-P-TB8 (X2)

AC adapter 6A-061WP12

17

4

5

C typeF typeG typeAC adapter（IP1500）

WDISR-IP500 WDISR-IP1500±0.5% (If 499 digit or less, ±3 digit.)(China)3

Accessories
Detector cable

Accuracy2
A type- Pack - Destina

Model

Dimensions

A typetype■ Equipped with RS232C and USB port

Range

1

Plug

Ordering number■ Easy-to-read color LCD display

Range

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.

18［p.41］

Torque Tester WDIS/WDISR-IPS series

Product specification detail

Automatic or semi automatic screw driver

Detector cable

Code

L1

L2

L4

1m（10P-1MF）

2m（10P-2MF）

4m（10P-4MF）

Contents

Package

Code

A

C

Attache case

Cardboard caseContents

Detector which is down sized for automation facility
Enable to choose the indicator to suit the usage scene

Destination

Power source can be selected from either AC adapter (WDIS) or internal battery (WDISR)

Model■ Various communication commands are available-

Detector■ Choice of power source and detector calble length&

SI unit (N･m)

SI unit (N･m)
Weight unit (kgf･cm)

cable

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

-

Pack
age

-

Destina

Plug

tion

type

■ Equipped with RS232C and USB port

1
A type2
A type(China)3

4

5

C typeF typeG typeAccessories

WDIS-IPS05C

WDIS-IPS5C

WDIS-IPS20CL

2.0〜500.0 ｍN･m

0.020~5.000 N･m

0.20〜20.00 N･m

OW joint

Cube joint

Detector cable

Rubber footAC adapter［p.42］［p.43］［p.47］［p.49］［p.41］±0.5% (If 499 digit or less, ±3 digit.)

Accuracy

AC adapter (or dry cell battery)* Tester power will be interlocked with external power source

Model

WDISR-IPS05C

WDISR-IPS5C

WDISR-IPS20CL

Measurement Range

2.0〜500.0 ｍN･m

0.020~5.000 N･m

0.20〜20.00 N･m

Accuracy±0.5% (If 499 digit or less, ±3 digit.)

PowerInternal battery (Ni-MH)

Dimensions(Used in outside of Japan)

Ordering number■ Prepare various measurement range

Power

Ox

(Used in Japan only)unit■ Easy-to-read color LCD display

Measurement Range

JxContents
measurement

Features

Model

Code

Indicator : 103(W) X 108(D) X 38(H) , Detector : See to p.57
Measurement joint OW-025

Measurement joint OW-10

Measurement joint OW-20

Cube joint SC-1

Cube joint SC-2

Cube joint SC-3

Accessories

Detector cable
Rubber foot B8-4
AC adapter 6A-061WP12* Each measurement joint comes with 2 spare screws.

1920Torque Tester
Electric impact tool

WIMS-IP series
Oil pulse tool

Air impact tool

Product specification detail

Detector cable

Torque wrench

Code

L1

L2

L4

0.9m（10P-1MF）

1.9m（10P-2MF）

3.9m（10P-4MF）

Contents

Package

Code

A

C

Attache case

Cardboard caseContents

Measuring sensitivity can be changed.

* For IP200 / 500 / 1500, only cardboard is available

It is possible to judge the passing of torque and number of blows.
WIMS series is a torque tester ideal for daily inspections of impact tool and torque wrench.

Destination

It is also possible to check the actual tightening torque of the torque wrench for operator training.

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Features

Ordering number■ Easy-to-understand color LCD.

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
C o d e■ Various commands for external control.

Plug type■ Measures the number of blows at
Model

the same time as the torque.

-■ Measurement sensitivity can be changed.

Detectorcable

1
A type- Package - Destination

2
A type(China)3

4

5

C typeF typeG typeAccessories■ Save up to 10 different setting values.

Detector cable■ Two types of communication ports (USB/RS232C).

Rubber footHandleAC adapter（IP1500）
Model

WIMS-IP50WIMS -IP200

WIMS-IP500

WIMS-IP1500

Except impact tool

1.00 〜 50.00 N・m

10.0 〜 200.0 N・m

10.0 〜 500.0 N・m

100 〜 1500 N・m

Impact tool (Note)1.00 〜 30.00 N・m

10.0 〜 120.0 N・m

10.0 〜 300.0 N・m

100 〜 600 N・m

Measurement Range［p.47］［p.49］

Socket adapter

Conversion socket［p.46］［p.46］［p.49］±0.5% (If 499 digit or less, ±3 digit.)

Accuracy
PowerInternal battery (Ni-MH)
Indicator︓103(W) X 108(D) X 38(H)Dimensions

Detector︓See to p.57
Socket adapter FNA-3D

Conversion socket

Conversion socket

NA34

NA46

Accessories

Conversion socket

NA48

Detector cable

HandleRubber foot B8-4

CPH-P-TB8 (X2)

AC adapter 6A-061WP12(Note) When using an impact tool, please make sure that the tightening torque value of used tool does notexceed the measurement range of the tester.

21

22［p.41］

Torque Tester DI-1M-IP series
Electric impact tool

Oil pulse tool

Air impact tool

Product specification detail
Torque wrench

Detector cable

Code

L1

L2

L4

1m（10P-1M）

2m（10P-2M）

4m（10P-4M）

Contents

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)Manage the blow force and number that affect tightening* x specifies the AC adapter plug.

DI-1M-IP series is a torque tester ideal for daily inspections of impact tool and torque wrench.

Plug

Code

1
A typetype

Accessories

Features
Ordering number■ Easy-to-see high brightness LED■ Simultaneously measure the blow force andModelblow number

DI-1M-IP50

AC adapter［p.48］［p.41］- Detector - Destinationcable■ Equipped with USB port
Model

Detector cable

DI-1M-IP200

DI-1M-IP500

Measurement range
Except impact tool
Impact tool
Accuracy

0.30~50.00 N･m

3.0~200.0 N･m

3.0~500.0 N･m

0.30~30.00 N･m ※3.0~120.0 N･m ※3.0~300.0 N･m ※±0.5% (If 499 digit or less, ±3 digit.)

PowerInternal battery (Ni-MH)

Dimensions

Indicator : 150(W) X 115(D) X 48(H)
Detector : See to p.57
Detector cable

Accessories

AC adapter 6A-061WP12(Note) When using an impact tool, please make sure that the tightening torque value of used tool does notexceed the measurement range of the tester.

23

24

2
A type(China)3

4

5

C typeF typeG typeIndex (Tightening tools, other)

0.2~120 N・m
0.002~4 N・m〇

0.02~1.5 N・m〇

0.0002~10 N・m〇

0.0015~15 N・m〇〇〇

0.002~2 N・m

25

100

Model

Page

RS232C

〇

10

USB

1

COM type

Dry cell battery

0.1

Internal battery

0.01

AC adapter

0.001

AC power

Friction torque inspection

Slip torque inspection

Tightening (Driver type)Tightening (Wrench type)

Power source

DIW series

27

〇

〇

△

DID series

29

〇

〇

△

NDID-150CN

31〇

WDIS-RL series〇

33

WDISR-RL series
DI-12 series

35

NTS7series

37

26

〇
〇

〇

△
〇

〇

〇
〇

〇

〇

Tightening tools (integrated type)

Product specification detail

Torque Wrench DIW series

Destination

Bolt tightening

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

1

Plug

A typetype

Accessories
One way drive

Ratchet adapter

AC adapter［p.46］［p.46］［p.41］

No hand force points for stable tightening operations
High operability to check tightening torque
DIW series is a digital torque wrench which is ideal for the bolt tightening .

Ordering number

Features■ No hand force pointModel■ Square drive parts is rotatable- Destination■ Equipped with pass/fail function

Original mechanism■ Equipped with the tightening
counter function

Easy to work due to check tightening■ Memory size is 800 data

torque by the rotating mechanism■ Equipped with USB port

Model
Measurement range

DIW-150.20〜15.00 N･m

DIW-20DIW-75

DIW-120

0.20〜20.00 N･m

0.20〜75.00 N･m

2.0〜120.0 N･m±0.5％ (If 199
Accuracy

digit or less,±1 digit.)±0.5％±0.5％ (If 199 digit or(If 499 digit or less, ±3 digit.)

less, ±1 digit.)

PowerInternal battery (Ni-MH)

Dimensions
Square drive
Accessories

See to p.56
6.35 sq.

9.5 sq.

9.5 sq.

One way drive

One way drive OWD-20

One way drive OWD-75

OWD-1512.7 sq.
Ratchet adapter 4755

AC adapter 6A-061WP12

27

28

2
A type(China)3

4

5

C typeF typeG typeTorque Driver DID series

Product specification detail

Screw tightening

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

1

Plug

A typetype

2
A type(China)3

4

5

C typeF typeG typeAccessories

Light and sound notification, ideal for screw tightening management
DID-05 has a built-in one way function / DID-4 comes with a one way bit holder

Features
Ordering number■ Equipped with pass/fail function■ Selectable measurement units

Model■ Equipped with the tightening counter function

Bit

Bit

(DID-05)(DID-4)［p.44,45］［p.44］

Bit holder［p.47］- Destination■ Memory size is 800 data
■ Equipped with USB port
Model
Measurement range

DID-05DID-4

2.0〜500.0 mN･m

0.020〜4.000 N･m

Accuracy±0.5％ (If 499 digit or less, ±3 digit.)

PowerInternal battery (Ni-MH)

Dimensions
SocketSee to p.56
Φ4 (HIOS shank)

HEX 6.35 (using bit holder)
Bit (#1) B35+1X50H

Bit (#1) V05+1X4X44

Accessories

Bit (#2) B35+2X50G

Bit (#1) D73+1X4X40

Bit holder 10-4141
One way bit holder OWB-4141

AC adapter 6A-061WP12

29

One way

AC adapter

bit holder

30［p.47］［p.41］

Torque Driver NDID-150CN

Product specification detail

Screw tightening

Destination

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code1

Plug type

A type2
A type(China)3

4

C typeF typeAccessories

Digital torque driver which is equipped the click mechanism

Bit

HEX wrench

USB flash drive

AC adapter［p.44］［p.47］［p.49］［p.41］

NDID-150CN is rotary-slip at a specified torque and displays the tightening torque by digital.

Features
Ordering number■ Grip rotates the specified torque■ Record “measurement date” andModel“measurement time” together- Destination■ Equipped with the tightening counter function■ Record to USB flash drive(400 records to the body)■ Light and sound notification
Model
Measurement range
AccuracyPower

NDID-150CN
0.020〜1.500 N･m±(1.0%+1digit)* Guaranteed in the range of 0.300 to 1.500 N･m
Internal battery (Ni-MH)

Length overall

205 mmSocketHEX 6.35
Bit (#1) B33+1X25.4H
Bit (#2) B33+2X25.4H

Accessories

HEX wrench TWH-16
USB flash drive UFD-P4GBKAC adapter 6A-061WP12

31

32

Tightening tools (separated type)

Product specification detail

Torque Tester WDIS/WDISR-RL series

Detector cable

CodeScrew tightening

L1

L2

L4

0.9m（10P-1MF）

1.9m（10P-2MF）

3.9m（10P-4MF）

Contents

Package

Code

A

C

Attache case

Cardboard caseContents

Ideal for tightening management of M1 ~ M6 screwPower source can be selected from either AC adapter (WDIS) or internal battery (WDISR)

Destination
Features■ Various communication commands are available■ Choice of power source and detector calble length

Model

Detector-cable■ Equipped with RS232C and USB port

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

Weight unit (kgf･cm)

unit- Pack - Destina
age

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code

tion

1

Plug

A typetype
Model
Measurement

WDIS-RL005
0.20〜50.00 ｍN･m

Range

*0.05ｍN･m incremeent

Accuracy±2.0％ or 0.3ｍN･m

Power
Model
Measurement

WDIS-RL05

WDIS-RL6

WDIS-RL10

2.0〜500.0 ｍN･m

0.020〜6.000 N･m

0.20〜10.00 N･m±0.5% (If 499 digit or less, ±3 digit.)

AC adapter (or dry cell battery) * Tester power will be interlocked with external power source
WDISR-RL005
0.20〜50.00 ｍN･m

Range

*0.05ｍN･m incremeent

Accuracy±2.0％ or 0.3ｍN･m

WDISR-RL05

WDISR-RL6

WDISR-RL10

2.0〜500.0 ｍN･m

0.020〜6.000 N･m

0.20〜10.00 N･m

Power
Dimensions
Socket

±0.5% (If 499 digit or less, ±3 digit.)
Internal battery (Ni-MH)

Bit

Bit

Bit holder(RL005／05)（RL6／10）［p.44］［p.44］［p.47］

Detector cable

Rubber footAC adapter［p.47］［p.49］［p.41］HEX 6.35 (using bit holder)
Bit (#1) B35+1X50H
Bit (#2) B35+2X50G

Bit (#0) D73+0X2.5X40

Bit holder 10-4141

Bit (#1) D73+1X4X40

One way bit holder OWB-4141
Auxiliary handle

Accessories

PSAGN13-100 (X2)
Detector cable
Rubber foot B8-4
AC adapter 6A-061WP12

33

2
A type(China)3

4

5

C typeF typeG typeAccessories

Indicator : 103(W) X 108(D) X 38(H) , Detector : See to p.58
Φ4 (HIOS shank)

SI unit (N･m)measurement

Ordering number■ Easy-to-read color LCD display

Code

34

One way
bit holder

Auxiliary handle（RL10）［p.47］［p.47］

Slip Torque Inspection
Slip torque tester DI-12 series

Product specification detail

Detector cableScrew, bolt inspection

Code

L1

L2

L4

1m（10P-1M）

2m（10P-2M）

4m（10P-4M）

1m（SL-7P-1M）

2m（SL-7P-2M）

4m（SL-7P-4M）

SL02
SL4

Contents

SL15

Destination

Ideal for inspection of tightened screws
DI-12 series inspect the tightening torque of screw based on the retightening torque method.

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Features■ Stable inspection due to motorized drive
Model■ Automatically stops when slip is detected-■ Equipped with USB port

Measurement
range
Accuracy

cable-

DestinaDI-12-SL15

1.5〜200.0 ｍN･m

0.030〜4.000 N･m

0.30〜15.00 N･m±2.0% (If 499 digit or less,±1.0% (If 499 digit or less, ±5 digit.)±10 digit.)

3
C typePlug typetion

DI-12-SL4

Accessories
Bit

Bit

Detector cable

Detector cable（SL02）（SL4）(SL02／SL4)(SL15)［p.45］［p.44］［p.48］［p.48］

Power cable

AC power

Length overall

Indicator : 166(W) X 142(D) X 48(H)Φ4
Bit (#0) V05+0X4X44

Accessories

Detector

1
A typeDI-12-SL02

Power
SocketCode

Ordering number■ Can estimate tightening torque value

Model

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.

HEX 6.35
Bit (#1) B35+1X50H
Bit (#2) B35+2X50G

Square drive 6.35－

Detector cable
Power cable PWC

3536［p.41］

Friction Torque Inspection
Product specification detail

Rotation friction torque tester NTS7 series
Friction torqueOpening torqueDestinationscrew tightening

Code

Jx

Ox

Contents(Used in Japan only)(Used in outside of Japan)&

SI unit (N･m)

SI unit (N･m)measurement

Weight unit (kgf･cm)

unit

Yard-pound unit (lbf･in)* x specifies the AC adapter plug.
Code1

Plug type

A type2
A type(China)3

5

C typeG typeAccessories

Inspectable with friction torque or rotation angle
NTS7 series is ideal for managing frictional torque of various rotating objects.

Bit

Bit

（RT05）（RT2）［p.44］［p.44］

Detector cable- L2 - A -■ Easy-to-read color LCD display

Model■ Various communication commands are available* To order HEX6.35 specification,■ Changeable rotation speed

Destination［p.48］

AC adapter

please see below.

(5 to 50 RPM, 5 RPM step)Model■ Can be tightened at constant torque- L2 - A -Destination+ H635［p.41］■ Tester power will be interlocked with external power source■ Equipped with RS232C and USB port

Measurement range

NTS7-RT05(+H635) (Note 1)

NTS7-RT05

2.0〜500.0 mN･mPower

AC adapter
Indicator : 103(W) X 108(D) X 38(H)Length overall

Detector : See to p.58.
Φ4 (HIOS shank)

HEX 6.35

Bit (#0) D73+0X2.5X40

Bit (#1) B35+1X50H

Bit (#1) D73+1X4X40

Accessories

NTS-RT2
0.020〜2.000 N･m±1.0% (If 499 digit or less, ±5 digit.)

Accuracy

SocketRubber footcable

Ordering number

Features

Model

External portBit (#2) B35+2X50G
Detector cable RT-12P-2M
External port cable EXT-5P-1M
Rubber foot B8-4
AC adapter SMI36-24

37

38［p.49］［p.49］

Memo

39

40

Accessories

Measurement joint OW series

AC adapter 6A-061WP12
Model

Model

6A-061WP12-1

6A-061WP12-2

6A-061WP12-3

6A-061WP12-4

6A-061WP12-5

A typeA type (China)

C type (Φ4)

F type (Φ4.8)

G typeOW-025

OW-10

OW-20

OW-60

MAX 6 N･m

Plug

Measurement range

AC 100~240V (50~60Hz)

Rated input

0.3A or less
DC 12V

OutputMAX 0.25 N･m

MAX 1 N･m

MAX 2 N･m

Joint shape⊕ M2.6

⊕ M3

⊕ M4

⊕ M6

Accessory

PAN, M2.6X5 (x2)

UPSET, M3X5 (X2)UPSET, M4X5 (X2)UPSET, M6X6 (X2)

WDI series

0.5A or less

Certification

PSE、FCC、CCC、BSMI、CE

PSE、CE、UKCA、KC

DI-9M series

Compliant models

NDI series

WDI series, DI-9M series, CD series, NDI series, DI-4B-25

WDIS／WDISR – IPS series

WDIS／WDISR - IP series
Compliant* OW joint comes with 2 spare screws.

WDIS／WDISR - IPS seriesmodels

WDIS／WDISR - RL series

Measurement joint SJ series

DI-1M series、WIMS-IP series
DIW series, DID series, NDID-150CN, WM2.4-DSUB-9S* As for the G plug, some models are not compatible.

Model

SJ-1.5

SJ-3

SJ-10

SJ-10K

SJ-50

SJ-200

MAX 0.3 N･m

MAX 1 N･m

MAX 1 N･m

MAX 5 N･m

MAX 20 N･m

HEX socket

HEX socket

HEX socket

HEX socket

Hexagonal

(5mm)

(5mm)

(6mm)

(6mm)

(17mm)HEX attachment

HEX attachment

AC adapter SMI36-24
Model

SMI36-24-1

SMI36-24-2

SMI36-24-3

SMI36-24-5

A typeA type (China)

C type (Φ4)

G typeMeasurement range

Plug

Joint shapem⊕ M2.5

AC 100~240V（50~60Hz）

Rated input

1.0A or lessAccessoryDC 24V

Output

1.5A or less

Certification

PSE、FCC、CCC、CE、UKCA

Compliant－

HS5-M2.6 (X3) HS5-M3 (x3)

HEX attachment HEX attachment

HS6-M3 (x3)

HS6-M4 (x3)

Joint cap

Joint cap

Joint cap

Joint cap

W-CAP-10

W-CAP-10

W-CAP-13

W-CAP-13

WDI series

Compliant models

CD series

NTS7 seriesmodelsMeasurement joint HEX14 series

Power cable PWC
Model

MAX 0.15 N･

Model

PWC-1

PWC-3

A typeC type (Φ4)

HEX14-H250

HEX14-S250

Plug

Measurement range
CompliantDI-12 seriesmodels41

Joint shape

MAX 25 N･m

MAX 25 N･mHexagonal (17mm)

Hexagonal (17mm)

Compliant models

DI-4B-25

42－

Cube joint SC series
Model

Bit H5BIT／H6BIT series

SC-1

SC-2

SC-3

Model

H5BIT-D73

H5BIT-D76

Model

M1／1.2／1.4／2／3

M2.6／3／4／5／6

M4／5／6／8／10

H5BIT-B35

H6BIT-B34

H6BIT-B35

H5BIT-SET(4-pack)

Tapped hole

H5BIT-B34

Bit point

WDI series
DI-9M series

CompliantInsertion partCD series

models

NDI series

Compliant

WDIS／WDISR - IPS series

measurement

HEX attachment HS series
Model

HS5-M2.6

HS5-M3SJ-3, SJ-10

joint
Model
HS6-M3

HS6-M4

H6BIT-D73

H6BIT-D76

Model

H6BIT-SET(4-pack)
Bit point
Joint shape⊕ M2.6

⊕ M3

⊕ M3

⊕ M4

Compliantmeasurement

SJ-3, SJ-10

SJ-10K, SJ-50

Insertion part

joint
Compliant

Joint cap W-CAP-10／13
Model

W-CAP-10

measurement

SJ-10K, SJ-50

joint

W-CAP-13

Bit B33／B35 series
Compliantmeasurement

SJ-3, SJ-10

SJ-10K, SJ-50

Model

B33+1X2.54H

B33+2X2.54H

B35+1X50H

B35+2X50G

Bit point⊕ No.1⊕ No.2⊕ No.1⊕ No.2

joint
Insertion part

Grease JT-GREASE
Model

Compliantmeasurement
joint

Compliant

JT-GREASE

DID-4、WDIS-RL6／RL10,WDISR-RL6／RL10,

NDID-150CN

modelsDI-12-SL4、NTS7-RT05(+H635)／RT2

Bit D73 series
SJ-1.5／3／10／10K／50／200

Model

D73+0X2.5X40

D73+1X4X40

HEX14-H250／S250

Bit point⊕ No.0⊕ No.1

Insertion part
Compliantmodels43

WDIS-RL005／RL05, WDISR-RL005／RL05,
DID-05, NTS7-RT05

44

Bit V05 series

Socket adapter

Model

V05+0X4X44

V05+1X4X44

Bit point⊕ No.0⊕ No.1

FNA-3D

Model

FNA-3D

Insertion part

Insertion part
CompliantmodelsSquare drive
DID-05, DI-12-SL02

9.5 sq.

Socket bit H17SCKTBIT-A20
Model

H17SCKTBIT-A20

Socket size

Hexagonal (17mm)

Compliant

WIMS-IP50

models

DI-1M-IP50

Conversion socket

NA series

Model

NA34

NA46

NA48

Square drive

12.7 sq.

19.0 sq.

25.4 sq.

Inlet size

9.5 sq.

12.7 sq.

12.7 sq.

Compliant

WIMS-IP200

WIMS-IP500

models

DI-1M-IP200

DI-1M-IP500

Insertion part

Compliantmeasurement

SJ-200, HEX14-H250, HEX14-S250

joint

HEX socket H17SOCKET-SQ9.5
Model

H17SOCKET-SQ9.5

One way drive OWD series
Model

Socket size

Hexagonal (17mm)

Inlet size

9.5 sq.

WIMS-IP1500

OWD-15OWD-20OWD-75

6.35 sq.

9.5 sq.

9.5 sq.

DIW-15, DID-4

DIW-20DIW-75Compliantmeasurement

SJ-200, HEX14-H250, HEX14-S250

Square drive
Inlet size

jointCompliantConversion adapter BAE-34
Model

BAE-34modelsRatchet adapter 4755
4755

Square drive

12.7 sq.

Inlet size

9.5 sq.

Model

Square drive

12.7 sq.

Inlet size

12.7 sq.

Compliantmodels45

DIW-12046

Bit holder 10-4141/One way bit holder OWB-4141
Model

10-4141

Detector cable 10P-XM series
Model

10P-1M

Cable length
Bit holder

Hexagonal (6.35mm)

Inlet size

6.35 sq.

Compliant

DIW-15, DID-4, WDIS-RL6／RL10

models10P-2M

10P-4M

2m

4m

OWB-41411m

Connector

Indicator sideshape

Detector side

Square type (10pin)
Round type (10pin)

Compliant models

DI-1M-IP series, DI-12-SL02／SL4

WDISR-RL6／RL10

Detector cable SL-7P-XM series

Auxiliary handle PSAGN13-100
Model

Model

SL-7P-1M

SL-7P-2M

SL-7P-4M

1m

2m

4m

PSAGN13-100

Handle length

100 mmCompliant

Cable length

WDIS-RL10、WDISR-RL10

modelsIndicator sideshape

Detector side

Round type (7pin)

Compliant models

HEX wrench TWH-16
Model

Connector

DI-12-SL15

Detector cable RT-12P-2M (A)TWH-16

Model

Size

RT-12P-2M

80 x 28 mm

Width across flat

5 mm
Cable length

Detector cable 10P-XMF series
Model

10P-1MF

2m

Connector

Indicator side

Square type (10pin) (With lock mechanism)shape

Detector side

Round type (12pin)

Compliant models
10P-2MF

NTS7 series

10P-4MF

External port cable EXT-5P-1M
Model
Cable length

0.9 m

Connector

Indicator sideshape

Detector side

Compliant models

1.9 m

EXT-5P-1M

3.9 m

Square type (10pin) *with ferrite core
Round type (10pin)
WDIS-IP／IPS／RL series, WDISR-IP／IPS／RL series, WIMS series

47

Cable length1mConnector shape

Square type (5pin)

Compliant models

NTS7 series

48

Options

Handle CPH-P-TB8
Model

Fixed stand BH-STD

CPH-P-TB8

Model

BH-STD
WDI seriesSizeCompliant models

Compliant40(W) X 48(H)Dimensions

160(W) x 116(H) x 55(D)Weight

Model

B8-4

size

Φ8 X 3.5

Around 1.2kg

Fixed stand LV-STD
Model

LV-STD
WDIS-IP50／IP200／IP500／IP1500

WDIS-IP／IPS／RL series、
WDISR-IP／IPS／RL series
NTS7 series, WIMS series

Compliant

WDISR-IP50／IP200／IP500／IP1500

models

WIMS-IP50／IP200／IP500／IP1500
DI-1M- IP50／IP200／IP500

Dimensions

USB cable USB-ECOM510

200(W) x 200(H) x 200(D)WeightModel

USB-ECOM510

Connector

A type – Mini B

Cable length

1m

Square drive

NDI series

CompliantCompliant models

CD series
NDI series

Rubber foot B8-4

Compliant models

DI-9M seriesmodels

WDIS-IP1500、WDISR-IP1500、 WIMS-IP1500

Around 9kg

Measurement attachment SA seriesModel

SA2-5.5

SA2-7

SA2-8

SA2-10

Width across flat

5.5 mm

7 mm

8 mm

10 mm

6.35 sq.
WDIS-IP5／WDISR-IP5

modelsUSB flash drive UFD-P4GBK
ModelInterfaceModel

SA3-12

SA3-13

SA3-14

SA3-17

SA3-19Width across flat

12 mm

13 mm

14 mm

17 mm

19 mm

UFD-P4GBK

Square drive

USB 2.0 Compatible

Compliant

9.5 sq.
WDIS-IP50／WDISR-IP50

modelsCompliant models

NDID-150CN

Model

SA4-22

SA4-24SA4-27

SA4-30

Width across flat

22 mm

24 mm

27 mm

30 mm

Square drive
Compliantmodels12.7 sq.
WDIS-IP200／WDISR-IP200／DWT-200

Model

SA6-32

SA6-36SA6-41

Width across flat

32 mm

36 mm

41 mm

Square drive
Compliantmodels49

19 sq.
WDIS-IP500／WDISR-IP500

50

HT series adapter

AHT series
Model

USB cable KU-SLAMB520BK

AHT-10

AHT-12Model

KU-SLAMB520BK

Connector
Dimensions

A

A type – Mini B

Cable length2m
WDI series, NDI series

Shaft
diameterCompliantmodels

Φ10mm

Φ12mm

NTS7-RT05

NTS7-RT2

A20BSQ2

WDIS-IP／IPS／RL seriesmodelsWDISR-IP／IPS／RL series
WIMS series, DIW series, DID series, NTS7 series

USB cable KU20-2K

Socket adapter A20BSQ series
ModelCompliant

Model
A20BSQ3

A20BSQ4

KU20-2K

Connector

A type –B type

Cable length2mCompliantInsertion part
Square drive

6.35 sq.

9.5 aq.

12.7 sq.

Compliant

WDIS-IP05／IP5

WDIS-IP50

WDIS-IP200

modelsWDISR-IP05／IP5WDISR-IP50

WDISR-IP200

CD series, DI-9M seriesmodels

DI-1M-IP series, DI-12 series

RS232C cable KR-LK2
Model

KR-LK2

Type

Cross

Cable length2m
WDI series

Compliant

WDIS-IP／IPS／RL seriesmodelsWDISR-IP／IPS／RL series
WIMS series, NTS7 series

RS232C cable KR-M1
Model

KR-M1

Type

Straight

Cable length1mCompliant

NDI seriesmodels51

52

Repair parts

Battery MH series／2HR series

Measurement joint OW series

Model

MH1100AA5BN.0
WDI series
DI-9M seriesNo

Compliant

OW-025models

PAN+2.6X5
1

UPSET+3X5

OW-20
UPSET+4X5

OW-60

CD series

CompliantDI-4B-25

modelsWDISR-IP／IPS／RL series

UPSET+6X6

WIMS series

Screw

DI-1M-IP series

for OW joint

OW-SPG025

2

OW-10

OW-SPG10OW-SPG20

OW-SPG60

Model

MH800AAA4GN.0

CompliantDID-4

models

DIW series

Model

MH600AAK4BN.0

Springfor OW joint

Compliant

DID-05modelsModel

Measurement joint SJ series
CompliantmodelsSJ-1.5

SJ-3

SJ-10

SJ-SHAFT1.5

SJ-SHAFT3

SJ-SHAFT10

2HR-AAAUC

Compliant

NDID-150CN

models
SJ-10K

SJ-50

SJ-SHAFT50

SJ-200SJ-SHAFT200

Shaft

Thumb screw TS-B-M10X15
Model

TS-B-M10X15

Compliant

DI-9M series

WDI series

for SJ joiint
modelsSJ-SPG1.5

SJ-SPG3SJ-SPG10SJ-SPG50

CD series
NDI series

SJ-SPG200（5 discs）

Springfor SJ joint

53

54

Customization

Dimensions

RS232C output

DIW series
Customization(Torque tester’s ordering number) + RS

code

DI-9M series
CD series
DI-4B-25

Compliant

DI-1M-IP seriesmodels

DIW series
DID-4
NDID-150CN

Note.

USB port is changed to RS232C port.

RS232C output (Baudrate 9600bps)
Customization(Torque tester’s ordering number) + RS9600

code
Compliant

DI-1M-IP seriesmodelsNote.

USB port is changed to RS232C port.

OW joint Screw diameter
Customization(OW joint’s model) + Mx

code

(Screw diameter)
OW joint’s model

Changeable screw diameter

OW-025M2, 2.5, 3, 4

Screw

OW-10

M2, 2.5(2.6), 3, 4, 5

diameter

OW-20

M2, 2.5(2.6), 3, 4, 5

OW-60

M3, 4, 5* Tapped hole is metric coarse screw threads.
Note.

Specifications of the product accessories cannot bechanged.
Screws are not included with custom products.

55

DID series56

IP detector

RL detector

Model

IP05 / 5

IP50

IP200

IP500

IP1500Model

RL005 / 05

RL6 / 10

Shapes(units : mm)

ShapesA

Detector size

Φ40mm

Φ50mm

Φ60mm

Φ80mm

Φ100mmB

Height41.2mm

63mm

79mm

98.5mm102.5mmC

Thickness

5mm

10mm

15.1mm

16mm

33mm

D

Flange size

Φ60mm

Φ90mm

Φ108mm

Φ140mmΦ170mmE

Hole position

Φ50mm

Φ74mm

Φ90mm

Φ116mm

Φ140mm

Φ4.2mm

Φ6.3mm

Φ8.3mmΦ10.5mm

Φ13mm(Every 120°)(Every 120°)(Every 120°)(Every 120°)(Every 60°)

F

Hole size

G

Socket6.35sq9.5sq

12.7sq19sq

25.4sqWeightAround 200g

Around 600g

Around 1.8kg

Around 3.4kg

Around 7.8kg

IPS detector
Model

IPS05C / 5C

IPS20CLRT detector
型

式

RT05

RT2

RT05(+H635)

形

状

A
B

先端外径

Φ 10.3 mm

Φ 12.0 mm

Φ 13.3 mm

先端長さ38.0 mm55.5 mm32.5 mmC

本体長さ

135.5 mm

168.5mm135.5 mm

D

本体外径

Φ 37.0 mm

Φ 42.0 mm

Φ 37.0 mm

Shapes(units : mm)

57

58

Please check our website for the latest information on each product.
* The contents of this catalog are subject to change without notice.

SUGISAKI METER CO., LTD.

Dealer

4-2-12, Shirahane, Ryugasaki-shi,
Ibaraki, 301-0841, Japan【URL】 https://www.cedar.co.jp【E-mail】 sales@cedar.co.jp

59