English

INSTRUCTION MANUALFORDIGITAL BLOOD PRESSURE MONITOR

CH-650̲1901.indb 1

CH-650EnglishFrançais
中国语Pyсский
EspañolPortuguês
한국어‫ﺍﻟﻌﺮﺑﻴﺔ‬1901

2019/06/03 14:04

Contents
3 - GENERAL REMARKS

EnglishCH-650̲1901.indb 2

5 - PRECAUTIONS FOR USE AND MAINTENANCE
10 - IDENTIFICATION OF PARTS
12 - LOADING THE BATTERIES
13 - REMOVING THE BATTERIES
14 - HOW TO WRAP THE WRIST CUFF
16 - TIPS FOR OBTAINING ACCURATE MEASUREMENTS
18 - MEASURING YOUR BLOOD PRESSURE
20 - CALLING UP THE DATA STORED IN MEMORY
22 - ERROR INDICATIONS
23 - ABOUT “BLOOD PRESSURE”
24 - BLOOD PRESSURE Q&A
25 - BEFORE REQUESTING REPAIRS OR TESTING
26 - SPECIFICATIONS

Ensure that you have all the following components

Blood pressure
monitor unit

Storage case(for monitor)
2 AAA batteries

Instruction
Manual- Eng 2 -2019/06/03 14:04

1. To deflate the cuff quickly, press the “START/STOP” switch.
2. The attached cuff is suitable for wrist circumferences of 13.5-21.5 cm.
3. When attaching the cuff, the body of a blood pressure monitor
should be placed on the inside of your wrist.
4. The monitor is not intended to be a diagnostic device, because it isintended for home use.
5. Self measurement is not therapy! In no event should you change
the dosage of your medication prescribed by your physician.
6. As preparation for measuring your blood pressure, you should
urinate and then remain relatively still for 10 to 15 minutes prior tomeasurement.
7. Exercising, eating, drinking, smoking, etc., prior to measurement
may affect the results.
8. Your blood pressure varies constantly throughout the day.
Measurements should be taken regularly at the same time each day.
9. DO NOT place too much emphasis on the results of one
measurement. Keep a continuing record of your blood pressure
variations. A complete picture can only be obtained from a large
number of readings.
10. Emotional stress tends to cause your blood pressure to rise.

EnglishGENERAL REMARKS
General remarks on blood pressure and blood pressure
measurement

To obtain correct blood pressure measurements

1. Take five or six deep breaths and then relax before measuring yourblood pressure. If you are tense when taking the measurement, youwill not get a valid reading.
2. Your blood pressure will be elevated if you are anxious or irritated,suffering from lack of sleep or constipation, or have just taken some
exercise or eaten a meal.
3. Wrap the cuff around your wrist snugly. (See page 14)
4. DO NOT measure your blood pressure after bathing or drinking.
5. If you feel the need to urinate, do so before measuring your bloodpressure.
6. Measure your blood pressure where the room temperature isaround 20°C. DO NOT measure your blood pressure when it is toocold (below 10°C) or too hot (above 40°C) in the room.
7. DO NOT try to measure your blood pressure immediately after
drinking coffee or tea or after smoking.
8. Measure your blood pressure when you are relaxed and still. Keepthe center of the cuff at the level of your heart and DO NOT moveyour arm or talk.
- Eng 3 -CH-650̲1901.indb 3

2019/06/03 14:04

GENERAL REMARKS

EnglishCH-650̲1901.indb 4

9. DO NOT measure blood pressure when the cuff has been on yourwrist for a few minutes or more. During this time, your wrist will havebuilt up a higher blood pressure and you will not get valid reading.
10. If the Body Movement lndicator or lrregular Heartbeat (IHB) lndicatoris displayed, relax and measure your blood pressure again. If thelrregular Heartbeat (IHB) Indicator appears frequently, you are advised
to consult a doctor about your health condition. (See page 19)
11. The figure below shows the blood pressure classifications under the
WHO standards.
SYSTOLIC (mmHg)
180

Grade 3 Hypertension (severe)
Grade 2 Hypertension (moderate)
Remarks:
The graph is not exact, but
may be used as a guidein understanding noninvasive blood pressure
measurements. The device
is only intended for use by
adults.

160
Grade 1 Hypertension (mild)
140
130
120

High-Normal
NormalOptimal
80 85 90

100

110

DIASTOLIC (mmHg)

Definitions and Classifications of Blood Pressure Levels
12. Intended Use• This device is noninvasive blood pressure monitor by
oscillometric method. This device can measure the systolicblood pressure (SYS), the diastolic blood pressure (DIA) and thepulse rate automatically.
• This device is intended to be used for checking personal healthcondition at home under the direction of a physician and is notintended to be a diagnostic device.
• This device is not intended for use with neonates or infant, andnot intended for automatic cycling measurement.

Measure your blood pressure at the same time each day.

* Your blood pressure changes all the time. This means that data
gathered over a long period has far more significance than datafrom just one measurement. For this reason, you must measure yourblood pressure on a daily basis. Ideally, you should measure yourblood pressure at the same time each day, wherever possible.

- Eng 4 -2019/06/03 14:04

1. If you suffer from heart disease, high blood pressure or othercirculatory disease, consult your physician before using the monitor.
2. If you feel abnormal pressure of the cuff during use or any otherirregularity, reduce the pressure by immediately switching off thepower or remove the batteries and consult the sales outlet wherepurchased.
3. If you think the measurement is abnormal or if measurement makes
you feel unwell, discontinue use and consult your physician.
4. Blood pressure measurement may not be possible for anyone witha weak pulse or arrhythmia.
5. Repeated blood pressure measurement may cause problems suchas congestion or swelling in some people.
6. Frequently repeated blood pressure measurements will not giveaccurate results. Allow an interval of about 1 minute between
measurements.
7. If you suffer from a severe problem with blood circulation in yourarms, consult your physician before using the monitor. Failure to doso could be hazardous to your health.
8. Measurement may not be possible for anyone with insufficient
blood flow to the area where measurements will be taken or who
suffers from a frequent irregular heartbeat. Consult your physician
for advice on whether to use the monitor.
9. DO NOT wrap the cuff around an injured wrist.
10. DO NOT wrap the cuff around a wrist in which a drip (intravenous
infusion) is inserted, or which is being used for blood transfusionas part of medical treatment. Doing so could result in an injury or aserious accident.
11. DO NOT use the monitor in the vicinity of flammable gases such as
those used for anaesthesia. Doing so could ignite the gases andcause an explosion.
12. DO NOT use the monitor in enriched oxygen environments suchas a hospital’s hyperbaric chamber or oxygen tent. Doing so could
ignite the oxygen and cause a fire.

CH-650̲1901.indb 5

English

PRECAUTIONS FOR USE AND MAINTENANCE
Precautions for use- Eng 5 -2019/06/03 14:04

PRECAUTIONS FOR USE AND MAINTENANCE

EnglishCH-650̲1901.indb 6

13. DO NOT use mobile phones near the monitor as this could result ina malfunction.
14. If you use a cardiac pacemaker, consult your physician before using
the monitor.
15. Be sure to use this unit only for measuring blood pressure. DO NOT
use it for any other purpose.
16. Be sure to use only our authorized parts and accessories. Parts andaccessories not approved for use with the device may damage theunit.
17. DO NOT use this unit on infants.
18. Blood pressure measurement may not be possible for anyone with
common arrhythmias such as atrial or ventricular premature beats
or atrial fibrillation.

Maintenance Precautions
1. DO NOT store the blood pressure monitor in locations exposed todirect sunlight, high temperatures (over 60°C), low temperatures(below -20°C), high relative humidity (over 95%) or excessive
amounts of dust.
Make sure to store the blood pressure monitor, where children,
pets and or pests are not there.
2. DO NOT drop the blood pressure monitor or subject it to othershocks or vibration.
3. Remove the batteries if the monitor will be left unused for a long
period.
4. DO NOT attempt to disassemble the monitor.
5. DO NOT bend the cuff excessively.
6. If the monitor is very dirty, wipe it clean with a cloth moistened withsterilizing alcohol or a neutral detergent. Then wipe it with a dry
cloth.
7. NEVER clean the blood pressure monitor with thinners or benzene,as this could damage the monitor.
8. To clean the cuff, wipe it with a moist cloth. Avoid hard rubbing, as
this will cause air leakages.

- Eng 6 -2019/06/03 14:04

Warning No modification of this device is allowed.
Symbols Explanation

CH-650̲1901.indb 7: The CE marking is meant to provide information to market
inspectors in the EU member countries.
: Type BF applied part
: Keep dry: Warning: Refer to instruction manual before use: Appliance compliance WEEE directive: Manufacturer

Warning

English

PRECAUTIONS FOR USE AND MAINTENANCE

The device has been tested and homologatedin accordance with EN60601-1-2 for EMC.
This does not guarantee in any way that thedevice will not be affected by electromagnetic
interference. Avoid using the device in highelectromagnetic environment.

- Eng 7 -2019/06/03 14:04

PRECAUTIONS FOR USE AND MAINTENANCE
Nature and frequency of maintenance:

EnglishCH-650̲1901.indb 8This product is designed for use over an extended period of time;
however, it is generally recommended that it be inspected every two
years to ensure proper function and performance.

Protect the nature environment:

Please help to protect natural environment by respecting national and/or local
recycling regulations when disposing of the battery and the product at the end of
their useful live.

WEEE MARK

If you want to dispose this product, do not mix with general household
waste. There is a separate collection systems for used electronics products inaccordance with legislation under the WEEE Directive (Directive 2012/19/EU)
and is effective only within European Union.

Information on Disposal in other Countries outside theEuropean Union.

If you wish to discard used batteries, please contact your local authorities or dealer
and ask for the correct method of disposal.

Note for the battery symbol.

The symbol might be used in combination with a chemical symbol. In this case it
complies with the requirement set by the Directive for the chemical involved.

Reference European standard:

The blood pressure measuring device corresponds to regulation EN60601-1,
EN1060-1, EN1060-3, EN1060-4.

Calibration:

The blood pressure measuring device is generally recommended to have themonitor inspected every two years to ensure correct functioning and accuracy.
Please contact a distributor.

- Eng 8 -2019/06/03 14:04

Name :
Address :Factory
Name :
Address :CITIZEN SYSTEMS JAPAN CO.,LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan

English

PRECAUTIONS FOR USE AND MAINTENANCE

CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen,
Guangdong, China

European Representative:

Name :
Address :EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague, The
Netherlands• For technical data, please contact our European
Representative

Made in China

CH-650̲1901.indb 9- Eng 9 -2019/06/03 14:04

IDENTIFICATION OF PARTS
EnglishCH-650̲1901.indb 10

Component names<Main Unit>

Display

MEMORY Switch
START/STOP
Switch Stand-by

Battery cover

Wrist cuff

- Eng 10 -2019/06/03 14:04

IDENTIFICATION OF PARTS

English<Display>

Systolic bloodpressure

Diastolic bloodpressure
Irregular Heartbeat(IHB) Indicator
Pulse

Body Movement Indicator

CH-650̲1901.indb 11

Battery level indicator
Appears when the batteries are running
See page 13)
low. (- Eng 11 -2019/06/03 14:04

LOADING THE BATTERIESEnglishCH-650̲1901.indb 12(Batteries supplied with the unit should be loaded in compartment
before you use the blood pressure monitor.)

1 Open the cover of battery compartment.

Battery cover
Pressing down on thebattery cover, slide thecover off in the direction of
the arrow.

2 Place batteries in the compartment.
Put the batteries paying attention to the positive and negative
terminal symbols
and .
* The battery goes in negative side first.

Insert the battery near
the display first.

- Eng 12 -2019/06/03 14:04

3 Close the battery cover.

EnglishLOADING THE BATTERIES

Slide the cover in the directionof the arrow. Push the cover on
until it clicks.

REMOVING THE BATTERIES• When the icon
is displayed or nothing appears in the display,replace the old batteries with two new ones. All two batteries should
be changed at once.
• Use only alkaline batteries.
• Do not use rechargeable batteries.
• Do not use alkaline (LR03) and manganese (R03) batteries together.
• Batteries included are for demonstration purposes only. Battery life
may be shorter than specified.
• When disposing of used batteries, comply with governmental
regulations or environmental public institution’s rules that apply inyour country/area.

CH-650̲1901.indb 13- Eng 13 -2019/06/03 14:04

HOW TO WRAP THE WRIST CUFF
EnglishCH-650̲1901.indb 14

Wrap the wrist cuff around your wrist properly. Improper wrapping of
the cuff may result in measuring errors.

1 Open the wrist cuff.
2 Wrap the cuff around your left wrist

so that the monitor is positioned on
the inside of your wrist.
• Wrap the wrist cuff on bare skin.
• The wrist cuff is designed to fit
wrists from 13.5 to 21.5 cm incircumference.

3 Loosely fit the wrist cuff bypulling the end of the cuff.

- Eng 14 -2019/06/03 14:04

4 Position the top edge of the cuff

approximately 1 to 2 cm away fromthe crease of your wrist.

EnglishHOW TO WRAP THE WRIST CUFF

5 Secure the wrist cuff with the hookand-loop fastener.

• If the wrist cuff is wrapped
too loose or tight, errors orincorrect readings may result.

You may use your right wrist to measure your blood pressure.
However, your blood pressure may differ by about 10mmHg dependingon whether measured on your left or your right wrist, so ideally you
should measure your blood pressure on the same wrist each time.
To obtain accurate measurements• Wrapping the cuff too loosely or too tightly may prevent correct
measurement.
• Sit up straight while taking the measurement.

CH-650̲1901.indb 15- Eng 15 -2019/06/03 14:04

TIPS FOR OBTAINING ACCURATE MEASUREMENTS
EnglishCH-650̲1901.indb 16Your blood pressure varies according to your posture, the time of day
and a range of other factors. Ideally, you should measure your bloodpressure in the same posture at the same time every day.

➀ Make sure that your posture is correct.

Sit up straight, take a deep breath and relax.

➁ Place your palm facing up andkeep your fingers relaxed.

➂ It is important to have thecuff level with your heart.

Place a folded towel or similar
under your arm.

- Eng 16 -2019/06/03 14:04

CH-650̲1901.indb 17

EnglishTIPS FOR OBTAINING ACCURATE MEASUREMENTS

To obtain correct measurements• Do not clench your fist tightly.

• Do not wrap the wrist cuff too loosely.

• Do not change the position of themonitor.

• Do not move your body or talk during
measurement- Eng 17 -2019/06/03 14:04

MEASURING YOUR BLOOD PRESSURE
EnglishCH-650̲1901.indb 18

1 Wrap the wrist cuff around your left wrist.
(

See page 14 “HOW TO WRAP THE WRIST CUFF”.)

2 Make sure that your posture is correct.
(

 ee page 16 “TIPS FOR OBTAINING ACCURATESMEASUREMENTS”.)

3 Press the “START/STOP” switch to startmeasurement.
• All digits displayed is an initial display,
indicating that the monitor is functioning
normal.
Themark will also be displayed but
this does not indicate that the battery
needs to be replaced.

is displayed when a pulse is detected.

- Eng 18 -2019/06/03 14:04

4 The measurement results are displayed.
Systolic bloodpressure

EnglishMEASURING YOUR BLOOD PRESSURE

Diastolic bloodpressure
Irregular Heartbeat(IHB) Indicator

CH-650̲1901.indb 19

Pulse
Body Movement Indicator

The Body Movement Indicator is displayed on themeasurement results display if you move your handor wrist during measurement, causing a substantial
pressure change to be detected.
• If the Body Movement Indicator is displayed,measure your blood pressure again.
If an irregular heartbeat is detected during
measurement, the Irregular Heartbeat (IHB) Indicator isdisplayed after measurement is completed.
An irregular heart beat is defined as a heart beatthat varies by ±25% from the middle of 5 heart beatintervals during the blood pressure measurement.
• Measurement may not be correct if your heartbeat
fluctuates greatly during measurement. If the
Irregular Heartbeat (IHB) Indicator is displayed,measure your blood pressure again.

CautionIf the Irregular Heartbeat (IHB) Indicatorappears frequently, you are advised toconsult a doctor about your health condition.

- Eng 19 -2019/06/03 14:04

MEASURING YOUR BLOOD PRESSURE

EnglishCH-650̲1901.indb 20

5 Finishing measurement.
• Press the “START/STOP” switch to turn the monitor off.

AUTOMATIC POWER OFF FEATURE• After taking the blood pressure measurement, pressing the “START/STOP” switch will turn the monitor off. However, if you do not press
the switch, the monitor’s Auto OFF function will automatically turnthe monitor off after 3 minutes.

CALLING UP THE DATA STORED IN MEMORY
60 measurement results can be stored in the memory.
The average value is calculated automatically to help you manage yourdaily health.

HOW TO CALL UP THE DATA
1 Press the “MEMORY” switch.

• “3A” indicating the average value appears on the left sectionof the display. Then, the average value, based on the last threemeasurements, is displayed. (“3A” and the average value do notappear if there are two or fewer measurements stored in memory.)- Eng 20 -2019/06/03 14:04

2 Each time you press the switch, the measurement data aredisplayed in order from the most recent to the oldest.
The data number
is displayed before
the measurement
results.

EnglishCALLING UP THE DATA STORED IN MEMORY

The measurement results are displayed after the data number.
• The value stored in memory is numbered in the order ofmeasurements. For example, when 60 sets of data are stored inmemory, the data number 1 represents the latest data.
Data number: 1, 2, 3, ... ... 59, 60ÇÇThe latest data
The earliest data

HOW TO DELETE THE DATA
1 Press the “MEMORY” switch.
2 While pressing the “MEMORY” switch, press and hold the“START/STOP” switch for more than 3 seconds.
All the stored data will be deleted.

CH-650̲1901.indb 21

Press andhold for 3
seconds

Deleted

A message
saying all thedata have been
deleted fromthe memory isdisplayed.

- Eng 21 -2019/06/03 14:04

ERROR INDICATIONSEnglishCH-650̲1901.indb 22

DISPLAY MARKCONDITION/CAUSECORRECTIVE ACTION

Appears when the wrist
cuff is wrapped too
loose.

Re-wrap the wrist cuff
snugly. Take anotherreading, making sure
you remain still. (See page 14 “HOWTO WRAP THE WRISTCUFF”.)

Appears when yourwrist moved during themeasurement.; orAppears when the cuff is
not wrapped correctly.

Re-wrap the wrist cuff
correctly. Take anotherreading, making sure
you remain perfectly
See page 14
still. (“HOW TO WRAP THE
WRIST CUFF”.)

Appears when thebatteries have run out.

Replace two alkaline
batteries with new
See pageones. (
13 “REMOVING THE
BATTERIES”.)

Appears when the pulse
rate falls outside themeasuring range (below
39 beats/min. or above
181 beats/min.)Re-wrap the wrist cuff
properly. Take deep
breaths to relax, andmeasure again. (See page 16 “TIPS FOROBTAINING ACCURATE
MEASUREMENT”.)

The unit does not
function properly.

Contact the store where
it was purchased or
your local service center.
See page 25(“BEFORE REQUESTINGREPAIRS OR TESTING”.)- Eng 22 -2019/06/03 14:04

ABOUT “BLOOD PRESSURE”
The heart is a pump that circulates blood throughout your body. Blood is pumped
from the heart at a constant pressure into arteries. This pressure is called thearterial blood pressure and represents, in general terms, your blood pressure. Bloodpressure is indicated by several kinds of pressures, including the systolic pressurethat occurs when the heart pumps blood, and the diastolic pressure that occurs
when blood returns to the heart.
(Systolic)(Diastolic)ArteryLeft atrium

Left
ventricle

EnglishWhat is blood pressure?

ArteryLeft atrium

The artery wall
expands.

Left
ventricle

Contraction

Normal state is restored tothe artery wall.
DilationYour blood pressure changes all the time.

Your blood pressure differs according to your age, gender and a range of otherfactors. It is likely to be affected by your biorhythm during the day and by yourposture, physical activities, mental activities, level of stress and even by the airtemperature. The blood pressure of a healthy person generally varies within a day.

[Example of blood pressure variations within a day]Systolic bloodpressure

Blood pressure value (mmHg)

160
140
120
100
80
60

Diastolic bloodpressure

40

Sleeping

12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11Time of day(Blood pressure values measured at five-minute intervals through the day)l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329–44.

CH-650̲1901.indb 23- Eng 23 -2019/06/03 14:04

BLOOD PRESSURE Q&A
EnglishCH-650̲1901.indb 24

Q Why is the blood pressure measured at home different from thatmeasured by the physician or at a hospital?

A Your blood pressure may vary due to exercise, the ambient temperatureor your mental state. When you have a physician or nurse measure yourblood pressure, the pressure value is likely to be 10 to 20 mmHg higher
than usual due to anxiety and/or stress. Knowing what affects your bloodpressure helps you to better monitor your health.

Q Why does the blood pressure obtained vary with measurement?
A Our blood pressure is adjusted by automatic nerve function. Bloodpressure differs with every heart beat. We are likely to think that our
blood pressure is constant, but it varies if you take measurements
in repetition. Blood pressure is susceptible to the time of day, suchas morning and afternoon, season and atmospheric temperature. In
addition, blood pressure is easily affected by mental stress or emotional
ups and downs. It tends to increase when you feel tension or decrease
when you are relaxed.

Q What is the benefit of measuring my blood pressure at home?
A Blood pressure measurements taken at home give reliable data as they

can be taken when you are in a stable and relaxed condition. Physicians
place considerable importance on blood pressure measurements taken
at home. You should measure your blood pressure at the same timeeach day and record the variations in your blood pressure.

Always consult your physician for an interpretation of yourblood pressure readings and to determine the proper
treatment.

- Eng 24 -2019/06/03 14:04

Problem

ChecksResponse

Display stays
blank when
you press the“START/STOP”
switch.

Check whether thebatteries are dead.

Replace all the batteries
with new ones.

Check whether thebatteries are installed
the right way round( and oriented
correctly).

Load the batteries in thecorrect direction.

Check whether themeasurement icon
is displayed.

Attach the cuff properly.

Nomeasurements
can be taken.

EnglishBEFORE REQUESTING REPAIRS OR TESTING
Make sure to check the following before sending your blood pressure
monitor away to be tested or repaired.

Check whether the
cuff is attachedcorrectly.
Did you stay
calm during
measurement?

Measure your bloodpressure again, making
sure to remain still.

Note that blood pressure measurement may not
be possible for someone with an extremely weakpulse or cardiac arrhythmia (irregular pulse).
Your blood pressure readings areabnormally high or low compared withthose taken in hospital.

Remove the batteries fromthe unit and change themwith new ones.

Other phenomena

CH-650̲1901.indb 25

See page 24.

- Eng 25 -2019/06/03 14:04

SPECIFICATIONSEnglishCH-650̲1901.indb 26

Model Number
Measurement system
Display
Wrist circumference range
Measurement Pressure
range
Pulse
Pressure
Accuracy
Pulse
Inflation
Rated voltage
ExhaustPower supply

CH-650
Oscillometric method
Digital display type13.5 to 21.5 cm
0 to 280 mmHg
40 to 180 pulse/min±3 mmHg±5% of reading
Automatic inflation by internal pump
3V DC(
: direct current)Electromagnetic quick exhaust valve
2 AAA batteries (LR03)(180 mmHg,
Battery duration
Approx. 400 times
once/day, 22°C)
OperatingTemperature 10°C to 40°C
Humidity
15% to 85% RH
conditions
Temperature -20°C to 60°C
Storageconditions
Humidity
10% to 95%RH
Memory Function
Up to 60 measurements
Electric shock Protection
Internal power unit
Degree of Protection
Type BF applied partMode of operation
Continuous operation
Protection against ingress of water IPX0
Dimensions
65(W) x 74(H) x 29(D) mm
WeightApprox. 103g w/o batteriesService Life
5 years
Set includes 2 AAA batteries (LR03) for theAccessories
monitor, Instruction Manual, Storage case
* Applied part for this device is Cuff.
* A range in barometric pressure 700 hPa to 1060 hPa*O verall system accuracy of this device is met with the requirement of item
7.9 of EN1060-3.
EN1060-3 : Non-invasive sphygmomanometers Part 3 (European Standard)- Eng 26 -2019/06/03 14:04

ELECTOROMAGNETIC COMPATIBILITY INFORMATION

inches) to any part of the [CH-650], including cables specified.
• Use of this equipment adjacent to or stacked with other equipment should be avoided.
• Use of accessories and options other than those specified (other than CITIZEN
original parts) could result in increased electromagnetic emissions or decreased
electromagnetic immunity of this equipment.
Guidance and manufacturer’s declaration - electromagnetic emissions
The [CH-650] is intended for use in the electromagnetic environment specified below. The
customer or the user of the [CH-650] should assure that it is used in such an environment.
Emissions test
ComplianceElectromagnetic environment - guidance
RF emissions
Group 1
The [CH-650] uses RF energy only for its internal
CISPR 11function. Therefore, its RF emissions are very
low and are not likely to cause any interference innearby electronic equipment.
RF emissions
Class BThe [CH-650] is suitable for use in all
CISPR 11establishments, including domestic
establishments and those directly connected toHarmonic emissions
N/A
the public low-voltage power supply network thatIEC 61000-3-2
supplies buildings used for domestic purposes.
Voltage fluctuations/ flicker N/Aemissions IEC 61000-3-3

EnglishWARNING • Portable RF communications equipment should be used no closer than 30 cm (12Guidance and manufacturer’s declaration - electromagnetic immunity
The [CH-650] is intended for use in the electromagnetic environment specified below. The
customer or the user of the [CH-650] should assure that it is used in such an environment.
Immunity test

IEC 60601
test level±8 kV contact
Electrostatic
discharge (ESD)
IEC 61000-4-2 ±15 kV air
Electrical fasttransient/burst
IEC 61000-4-4
Surge
IEC 61000-4-5
Voltage dips,
shortinterruptionsand voltage
variations on
power supply
IEC 61000-4-11

Compliance
level

Electromagnetic environment guidance±8 kV contact

Floors should be wood, concrete or
ceramic tile. If floors are covered withsynthetic material, the relative humidity
should be at least 30%.
N/A

±15 kV air±2 kV for
N/A
power supplylines±1 kV for
input/output lines
±1 kV
N/A
line to line
±2 kV
line to earth
0% UT
0.5 cycle

N/A

N/A

N/A

0% UT
1 cycle70% UT
25/30 cycle
0% UT
250/300 cycle

Power frequency(50/60 Hz)
30 A/mmagnetic field
IEC 61000-4-8

30 A/mPower frequency magnetic fields should
be at levels characteristic of a typicallocation in a typical commercial or
hospital environment.

Note: UT is the A.C. mains voltage prior to application of the test level.

CH-650̲1901.indb 27- Eng 27 -2019/06/03 14:04

EnglishCH-650̲1901.indb 28

Guidance and manufacturer’s declaration - electromagnetic immunity
The [CH-650] is intended for use in the electromagnetic environment specified below. The customer or theuser of the [CH-650] should assure that it is used in such an environment.
Portable and mobile RF communications equipment should be used no closer to any
part of the [CH-650], including cables, than the recommended separation distancecalculated from the equation applicable to the frequency of the transmitter.
Recommended separation distance
N/A
Conducted 3 Vrms, 6 Vrms N/A
150 kHz
RF
to
IEC
61000-4-6 80 MHz
Radiated
RF
IEC
61000-4-3

10 V/m 10 V/m d = 1.2
80 MHz to 800 MHz
80 MHzd = 2.3
800 MHz to 2.7 GHz
to
2.7 GHz
where P is the maximum output power rating of the transmitterin watts (W) according to the transmitter manufacturer and d isthe recommended separation distance in meters (m).
Field strengths from fixed RF transmitters, as determinedby an electromagnetic site survey,a) should be less than thecompliance level in each frequency range.b)Interference may occur in the vicinity ofequipment marked with the following symbol:
NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation isaffected by absorption and reflection from structures, objects and people.
a). F ield strengths from fixed transmitters, such as base stations for radio (cellular/ cordless) telephones
and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be
predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF
transmitters, an electromagnetic site survey should be considered. If the measured field strength in thelocation in which the [CH-650] is used exceeds the applicable RF compliance level above, the [CH-650]
should be observed to verify normal operation. If abnormal performance is observed, additional
measures may be necessary, such as reorienting or relocating the [CH-650].
b). O ver the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.
Recommended separation distances between portable and mobile RF communications
equipment and the [CH-650]
The [CH-650] is intended for use in an electromagnetic environment in which radiated RF
disturbances are controlled. The customer or the user of the [CH-650] can help prevent
electromagnetic interference by maintaining a minimum distance between portable and mobile
RF communications equipment (transmitters) and [CH-650] as recommended below, according
to the maximum output power of the communications equipment.
Rated maximum
Separation distance according to frequency of transmitter (m)output power of 150 kHz to 80 MHz 150 kHz to 80 MHz 80 MHz to 800 MHz 800 MHz to 2.7 GHztransmitterd = 1.2
d = 2.3
N/A
N/A
(W)
0.01
N/A
N/A
0.12
0.23
0.1
N/A
N/A
0.38
0.73
1
N/A
N/A
1.2
2.3
10
N/A
N/A
3.8
7.3
100
N/A
N/A
12
23
For transmitters rated at a maximum output power not listed above, the recommended separation distance din meters (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is
the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected byabsorption and reflection from structures, objects and people.

- Eng 28 -2019/06/03 14:04

CH-650̲1901.indb 29

EnglishGuidance and manufacturer’s declaration - electromagnetic immunity
The [CH-650] is intended for use in the electromagnetic environment specified below. The
customer or the user of the [CH-650] should assure that it is used in such an environment.
Immunity to
Test
Band a) Service a)
Modulation
Maximum Distance IMMUNITY
proximity fields Frequncy (MHz)
b)
power
(m)
TEST LEVEL
from RF wireless (MHz)
(W)
(V/m)
communications 385
380-390 TETRA 400
Pulse
1.8
0.3
27
equipment IEC
modulation
61000-4-3b) 18 Hz450
430-470 GMRS 460,
FM c)
2
0.3
28
FRS 460± 5 kHzdeviation
1 kHz sine
710
704-787 LTE Band
Pulse
0.2
0.3
9
13,
17
modulation
745b) 217 Hz
780
2
0.3
28
810
800-960 GSM 800/900, Pulse
modulation
TETRA 800,
870
iDEN 820,b) 18 Hz930
CDMA 850,
LTE Band 5
1720
1700GSM 1800;
Pulse
2
0.3
28
1900
CDMA 1900; modulation
1845
GSM 1900;b) 217 Hz
1970
DECT;
LTE Band 1, 3,
4, 25; UMTS
Pulse
2
0.3
28
2450
2400Bluetooth,modulation
2570
WLAN,
802.11 b/g/n, b) 217 Hz
RFID 2450,
LTE Band 7
0.2
0.3
9
5240
5100WLAN 802.11 Pulse
a/n
modulation
5800
5500
b) 217 Hz
5785
NOTE If necessary to achieve the IMMUNITY TEST LEVEL, the distance between thetransmitting antenna and the [CH-650] may be reduced to 1 m. The 1 m test distance ispermitted by IEC 61000-4-3.
a). F or some services, only the uplink frequencies are included.
b). The carrier shall be modulated using a 50 % duty cycle square wave signal.
c). As an alternative to FM modulation, 50 % pulse modulation at 18 Hz may be used because
while it does not represent actual modulation, it would be worst case.

- Eng 29 -2019/06/03 14:04

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 30

2019/06/03 14:04

CH-650̲1901.indb 1

Français

MANUEL D’UTILISATIONPOUR
L’AUTO-TENSIOMÈTRE NUMÉRIQUE

CH-650Français

1901

2019/06/03 14:04

Table des matières
3 - GÉNÉRALITÉS
5 - PRÉCAUTIONS D’EMPLOI ET ENTRETIEN
10 - NOMES DES PIÈCES ET LEUR EMPLACEMENT

FrançaisCH-650̲1901.indb 2

12 - MISE EN PLACE DES PILES
13 - RETRAIT DES PILES
14 - FIXATION DU BRASSARD
16 - ASTUCES POUR OBTENIR DES MESURES DE PRECISION
18 - MESURE DE LA TENSION
20 - RAPPEL DES DONNÉES MÉMORISÉES
22 - INDICATIONS D’ERREUR
23 - À PROPOS DE LA «TENSION ARTÉRIELLE»
24 - QUESTIONS ET RÉPONSES SUR LA TENSION
25 - AVANT DE SOLLICITER UNE RÉPARATION OU DES TESTS
26 - SPÉCIFICATIONS

Vérifiez que toutes les pièces suivantes sont fournies

Auto-tensiomètreBoîtier derangement(pour l’appareil)
2 piles AAA

Manueld’utilisation- Fra 2 -2019/06/03 14:04

GÉNÉRALITÉS
1. Pour dégonfler le brassard rapidement, appuyez sur le bouton«START/STOP».
2. Le bracelet de poignet fourni s’adapte à une circonférence de poignet
de 13,5 à 21,5 cm.
3. Lorsque vous fixez le brassard, la structure d’un appareil de surveillancede la pension devrait être placé à l’intérieur de votre poignet.
4. Le moniteur n’est pas conçu pour servir d’appareil de diagnostic, caril est destiné à un usage privé.
5. L’auto-mesure de la tension n’est pas une thérapie ! Vous ne devez en
aucun cas modifier la dose des médicaments prescrits par votre médecin.
6. En préparation de la mesure de la tension, il est recommandé d’uriner,
puis de rester relativement immobile pendant 10 à 15 minutes avantde réaliser la mesure.
7. Tout exercice, toute consommation d’aliments ou de boisson, ou
toute cigarette fumée avant la mesure risque d’influencer les résultats.
8. Votre tension varie en permanence au cours de la journée. La mesure
doit être réalisée régulièrement et chaque jour à la même heure.
9. N’accordez PAS trop d’importance aux résultats d’une seule mesure.
Consignez les écarts de tension. Un aperçu complet ne peut êtreobtenu qu’à partir d’un grand nombre de lectures.
10. Un état de tension émotionnelle a tendance à provoquer une
augmentation de la tension.

Français

Remarques générales sur la tension et sur la mesure de la tension

Pour obtenir une mesure correcte de la tension

1. Respirez profondément cinq à six fois puis détendez-vous avant demesurer votre tension. Si vous êtes tendu en réalisant cette mesure,la valeur obtenue ne sera pas valide.
2. La tension est élevée lorsque vous êtes dans un état d’inquiétude ou
d’irritation, si vous manquez de sommeil ou si vous êtes constipé, ou
encore après une activité physique ou un repas.
Voir page 14)
3. Serrez le brassard autour de votre poignet. (4. NE mesurez PAS votre tension après un bain ou après touteconsommation de liquide.
5. Si vous sentez le besoin d’uriner, faites-le avant de mesurer votre tension.
6. Mesurez votre tension dans une pièce où la température est de 20°C
environ. NE mesurez PAS votre tension lorsque la température ambiante
est trop basse (inférieure à 10°C) ou trop élevée (supérieure à 40°C).
7. N’essayez PAS de mesurer votre tension immédiatement après avoir
consommé du thé ou du café ou après avoir fumé.
8. Mesurez votre tension lorsque vous êtes détendu et immobile.
Maintenez le centre du brassard au niveau de votre cœur, et NE bougez
PAS votre bras ou ne parlez pas.

CH-650̲1901.indb 3- Fra 3 -2019/06/03 14:04

GÉNÉRALITÉSFrançaisCH-650̲1901.indb 4

9. NE mesurez PAS la tension alors que le bracelet de poignet est en place
depuis quelques minutes, car votre poignet aura accumulé une tension
plus élevée, et la mesure ainsi obtenue sera incorrecte.
10. Si l’indicateur de mouvements corporels s’affiche, mesurez à nouveau
votre tension artérielle. Si l’indicateur de battements de cœur irréguliers
apparaît fréquemment, il est conseillé de consulter un docteur à proposde votre état de santé. (Voir page 19)
11. L’illustration ci-dessous présente la classification de la tension en
fonction des normes de l’OMS.
SYSTOLIQUE (mmHg)

180

Hypertension de la catégorie 3 (grave)
Hypertension de la catégorie 2 (modérée)160

Remarques :Le graphique n’est pas exact,
mais il peut aider à comprendre
les mesures non invasives dela tension artérielle. L’appareil
est destiné à être utilisé sur une
population adulte uniquement.

Hypertension de la catégorie 1 (douce)
140
130
120

Haut-Normale
NormalOptimums
80 85 90

100

110

DIASTOLIQUE (mmHg)

Définition et classification des niveaux de tension
12. Utilisation prévue• Ce dispositif est moniteur de pression artérielle non effractif
par méthode oscillométrique. Ce dispositif permet de mesurer
la pression artérielle systolique (SYS), la pression artérielle
diastolique (DIA) et la fréquence du pouls automatiquement.
•C e dispositif est destiné à être utilisé pour vérifier un état desanté personnel à la maison, sous la direction d’un médecin etn’est pas destiné à être un dispositif de diagnostic.
•C e dispositif n’est pas destiné à être utilisé avec un nouveau-né
ou un nourrisson, et ni à la mesure automatique de cycles.

Mesurez votre tension chaque jour à la même heure.

* Votre tension change en permanence. Cela signifie que les données
recueillies sur une longue période ont une plus grande signification quecelles issues d’une seule mesure. Il est donc nécessaire de mesurer
votre tension quotidiennement. Dans l’idéal, vous devriez mesurer votre
tension chaque jour à la même heure si possible.

- Fra 4 -2019/06/03 14:04

PRÉCAUTIONS D’EMPLOI ET ENTRETIEN
1. Si vous souffrez d’une maladie cardiaque, si vous faites del’hypertension ou si vous souffrez de toute autre maladie de lacirculation, veuillez consulter un médecin avant d’utiliser cet
appareil.
2. Si vous sentez que le brassard exerce une pression anormale
pendant l’utilisation ou si vous remarquez toute autre irrégularité,
diminuer immédiatement la pression en remontant le bouton Powerou retirez les piles puis consultez le point de vente où vous avez
effectué votre achat.
3. Si vous pensez que la mesure est anormale ou si le fait de mesurer
votre tension provoque chez vous un malaise, arrêtez la mesure etconsultez votre médecin.
4. La mesure de la tension peut s’avérer impossible pour une
personne ayant un pouls faible ou souffrant d’arythmie.
5. En cas de mesures répétées de la tension, une congestion, un
gonflement, etc. peut se produire chez certaines personnes.
6. En cas de mesures répétées de la tension, aucun résultat précis nesera obtenu. Un intervalle de 1 minute entre deux mesures doit êtrerespecté.
7. Consultez votre médecin avant d’utiliser cet appareil si voussouffrez d’un trouble grave de la circulation sanguine dans les bras.
Vous risqueriez autrement de mettre votre santé en danger.
8. La mesure peut ne pas être possible pour une personne ayant un
flux sanguin insuffisant dans la zone de mesure ou des battementsde cœur irréguliers. Consultez votre médecin pour obtenir son avis
sur l’utilisation de cet appareil.
9. NE placez PAS le bracelet de poignet autour d’un poignet blessé.
10. NE placez PAS le bracelet de poignet sur un poignet dans lequelune perfusion a été insérée (injection intraveineuse) ou s’il est
utilisé pour une transfusion sanguine dans le cadre d’un traitement
médical. Le non-respect de cette consigne risque de provoquer
une blessure ou un accident grave.
11. N’utilisez PAS cet appareil à proximité de gaz inflammables commeles gaz anesthésiants en raison du risque d’incendie et d’explosion.

CH-650̲1901.indb 5

Français

Précautions d’emploi- Fra 5 -2019/06/03 14:04

PRÉCAUTIONS D’EMPLOI ET ENTRETIEN

FrançaisCH-650̲1901.indb 6

12. N’utilisez PAS cet appareil dans des environnements enrichis en
oxygène comme la chambre hyperbare d’un hôpital ou une tente à
oxygène en raison des risques d’incendie.
13. N’utilisez PAS de téléphone portable à proximité de cet appareil
sous peine de provoquer un dysfonctionnement.
14. Si vous utilisez un stimulateur cardiaque, veuillez consulter votre
médecin avant d’utiliser cet appareil.
15. N’utilisez cet appareil que pour mesurer la tension. NE l’utilisez PASà d’autres fins.
16. Veillez à n’utiliser que nos pièces et accessories autorisés. Les
pièces et accessoires dont l’utilisation n’est pas approuvée avec
l’appareil peuvent endommager ce dernier.
17. N’utilisez PAS cet appareil sur un nourrisson.
18. La mesure de la tension artérielle peut s’avérer impossible pour despersonnes souffrant d’arythmies courantes comme une extrasystoleatriale ou ventriculaire, ou une fibrillation atriale.

Précautions d’entretien

1. NE conservez PAS le tensiomètre dans un endroit exposé à lalumière directe du soleil, à des températures élevées (supérieures à60°C), à des températures basses (inférieures à -20°C), à un tauxd’humidité relativement élevé (supérieure à 95%) ou dans un endroit
poussiéreux.
Assurez-vous de ranger l’auto-tensiomètre hors de la portée desenfants et des animaux domestiques.
2. NE le faites PAS tomber et ne le soumettez à aucune force ni
vibration.
3. Retirez les piles pendant les périodes d’inutilisation prolongée.
4. NE démontez PAS l’appareil.
5. NE pliez PAS le bracelet de poignet de façon excessive.
6. Lorsque le tensiomètre est très sale, nettoyez-le à l’aide d’un chiffon
humide imbibé d’alcool stérilisé ou d’un détergeant neutre, puis
essuyez-le à l’aide d’un chiffon sec.
7. Ne nettoyez JAMAIS le tensiomètre avec du diluant ou du benzène
sous peine de le détériorer.
8. Nettoyez le brassard avec un linge humide. Évitez de frotter trop
fort sous peine de provoquer des fuites d’air.

- Fra 6 -2019/06/03 14:04

PRÉCAUTIONS D’EMPLOI ET ENTRETIEN
modification de cet équipement n’est
Avertissement Aucune
autorisée.

CH-650̲1901.indb 7: Le marquage CE est destiné à fournir des informations
aux inspecteurs de marché dans les pays membres del’UE.
: Partie appliquée de type BF

Français

Explication des symboles: Garder au sec: Avertissement: Voir la notice technique avant utilisation: Directive WEEE sur la conformité des équipementsélectriques: Fabricant

Avertissement

Cet appareil a été testé et homologué
conformément à la norme EN60601-1-2 surla compatibilité électromagnétique. Ceci neconstitue en aucun cas une garantie que
cet appareil ne sera pas affecté par desinterférences électromagnétiques. Evitez
d’utiliser cet appareil dans un environnement
hautement électromagnétique.

- Fra 7 -2019/06/03 14:04

PRÉCAUTIONS D’EMPLOI ET ENTRETIEN
Nature et fréquence de l’entretien :Cet appareil a été conçu pour une utilisation pendant une période prolongée ; il est
toutefois généralement recommandé de l’inspecter tous les deux ans afin de vérifier
son fonctionnement et ses performances.

Protection de l’environnement :FrançaisCH-650̲1901.indb 8

Merci d’aider à protéger l’environnement en respectant les réglementations
nationales et/ou locales de recyclage lorsque vous mettez au rebut la pile et leproduit en fin de vie.

MARQUAGE WEEE

Si vous souhaitez vous débarrasser de cet appareil, ne le mettez
pas à la poubelle avec vos ordures ménagères. II existe un systèmede récupération distinct pour les vieux appareils électroniques
conformément à la législation WEEE sur le recyclage des déchets deséquipements électriques et électroniques (Directive 2012/19/EU) qui est
uniquement valable dans les pays de l’Union européenne.

Informations sur la mise au rebut dans les autres pays en
dehors de l’Union Européenne.

Si vous souhaitez mettre au rebut les piles usées, veuillez contacter vos autorités
locales ou votre revendeur pour la méthode de mise au rebut correcte.

Remarque sur le symbole des piles.

Le symbole peut être utilisé en combinaison avec un symbole chimique. Dans ce
cas, il est conforme avec les prescriptions mise en place par les directives de priseen charge des produits chimiques.

Norme de référence européenne :Le dispositif de mesure de la pression artérielle correspond à la réglementation
EN60601-1, EN1060-1, EN1060-3, EN1060-4.

Calibrage :Il est généralement recommandé d’inspecter l’appareil de mesure de lapression artérielle tous les deux ans afin de vérifier son bon fonctionnement et
sa précision. Veuillez contacter un distributeur.

- Fra 8 -2019/06/03 14:04

PRÉCAUTIONS D’EMPLOI ET ENTRETIEN

Usine

Nom :
Adresse :CITIZEN SYSTEMS JAPAN CO.,LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan
CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen,
Guangdong, China

FrançaisNom :Adresse :Représentant en Europe:
Nom :Adresse :EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague,
The Netherlands• Pour les données techniques, veuillez contacter notre
représentant en Europe.

Fabriqué en Chine

CH-650̲1901.indb 9- Fra 9 -2019/06/03 14:04

NOMES DES PIÈCES ET LEUR EMPLACEMENT
Noms des pièces<Unité principale>
FrançaisCH-650̲1901.indb 10

Affichage

Bouton MEMORY

BoutonSTART/STOP
VeilleCouvercle ducompartimentà piles

Bracelet de poignet- Fra 10 -2019/06/03 14:04

NOMES DES PIÈCES ET LEUR EMPLACEMENT

Pression systolique

Français<Affichage>

Pression diastolique

Indicateur depulsations
irrégulières (IHB)
Pulsation

Indicateur de mouvement
du corps

CH-650̲1901.indb 11

Indicateur de niveau des piles
Apparaît quand le niveau des piles est
Voir page 13)
faible. (- Fra 11 -2019/06/03 14:04

MISE EN PLACE DES PILES(Les piles fournies avec l’unité doivent être chargées dans lecompartiment avant l’utilisation du sphygmomanomètre.)

FrançaisCH-650̲1901.indb 12

1 Ouvrez le couvercle du compartiment à piles.

Couvercle ducompartiment à piles

Poussez sur le couvercle
du compartiment à piles
et faites-le glisser dans la
direction de la flèche.

2 Placez les piles dans le compartiment.
Placez-les piles en respectant le sens des polarités / .
* Placez la pile à l’intérieur en faisant entrer le côté négatif
premier.

en

Insérez d’abord la pileprès de l’affichage.

- Fra 12 -2019/06/03 14:04

MISE EN PLACE DES PILES

Faites glisser le couvercle dansla direction de la flèche. Puis
poussez sur le couvercle jusqu’à
ce que vous entendiez un clic.

Français

3 Fermez le couvercle du compartiment à piles.

RETRAIT DES PILES• Lorsque l’icône
apparaît ou lorsque l’écran est vierge, remplacez
les piles usagées par deux nouvelles piles. Il est important dechanger les deux piles simultanément.
• N’utilisez que des piles alcalines.
• N’utilisez pas de piles rechargeables.
• N’utilisez pas des piles alcalines (LR03) et manganèses (R03)
ensemble.
• Les piles fournies le sont uniquement dans un but de démonstration.
La durée de vie des piles peut être plus courte que la durée indiquée.
• Lorsque vous jetez des piles usagées, veuillez vous plier aux
règlements nationaux ou aux règles en matière d’environnement envigueur dans votre pays ou votre région.

CH-650̲1901.indb 13- Fra 13 -2019/06/03 14:04

FIXATION DU BRASSARD
Fixez correctement le brassard autour du poignet. Une mauvaise
fixation peut entraîner des erreurs de mesures.

FrançaisCH-650̲1901.indb 14

1 Ouvrez le brassard.
2 Attachez le brassard autour

du poignet de sorte que lesphygmomanomètre se trouve àl’intérieur du poignet.
• Fixez le brassard autour du poignet
nu.
• Ce brassard est conçu pours’adapter à des poignets d’une
circonférence de 13,5 à 21,5 cm.

3 Fixez sans serrer lebrassard en tirant sur son
extrémité.

- Fra 14 -2019/06/03 14:04

FIXATION DU BRASSARD

4 Placez l’extrémité supérieure duFrançaisbrassard à une distance d’environ
1 à 2 cm de la naissance du poignet.

5 Fixez le bracelet de poignet avec labande velcro.

• Si le brassard n’est pas
suffisamment serré, ou s’il
est trop serré au contraire,cela risque d’entraîner deserreurs ou des mesuresincorrectes.

Vous pouvez utiliser votre poignet droit pour mesurer votre
pression artérielle.
Cependant, votre pression artérielle peut différer d’environ 10 mmHg
selon que vous faites la mesure le poignet gauche ou sur le poignet
droit, c’est pourquoi il est préférable de mesurer votre pression artérielle
toujours sur le même poignet.
Pour obtenir une mesure précise
• Si vous ne serrez pas assez le bracelet de poignet ou si vous leserrez trop, une mesure correcte ne pourra pas être obtenue.
• Asseyez vous bien droit lors de la mesure.

CH-650̲1901.indb 15- Fra 15 -2019/06/03 14:04

ASTUCES POUR OBTENIR DES MESURES DE PRECISION
Votre tension varie en fonction de votre posture, l’heure de la journée etune variété d’autres facteurs. Dans l’idéal, vous devriez mesurer votre
tension dans la même posture chaque jour à la même heure.

FrançaisCH-650̲1901.indb 16➀ Assurez-vous que votre position est correcte.

Asseyez-vous bien droit, respirez un grand coup etdétendez-vous.

➁ Placez la paume de votre mainvers le haut et détendez vos
doigts.

➂ Il est important que lebracelet de poignet setrouve au même niveau
que votre cœur.

Placez une serviette pliée ou
quelque chose de similaire sous
votre bras.

- Fra 16 -2019/06/03 14:04

ASTUCES POUR OBTENIR DES MESURES DE PRECISION

CH-650̲1901.indb 17• Ne serrez pas votre poing trop fort.

Français

Pour obtenir des mesures correctes• Ne serrez pas le bracelet de poignet
trop fort.

• Ne changez pas la position del’appareil.

• Ne bougez pas votre corps et ne parlez
pas pendant la mesure.

- Fra 17 -2019/06/03 14:04

MESURE DE LA TENSION
1 Fixez le brassard autour du poignet gauche.
(

Voir page 14 «FIXATION DU BRASSARD».)

2 Vérifiez si la position est correcte.
FrançaisCH-650̲1901.indb 18(

Voir page 16 «ASTUCES POUR OBTENIR DES MESURES DEPRECISION».)

3 Appuyez sur le bouton «START/STOP»

pour démarrer la mesure.
• Tous les chiffres affichés représentent
l’affichage initial et indiquent que l’appareil
fonctionne normalement.
La marque
est aussi affichée maiscela ne signifie pas que la pile doit êtreremplacée.

est affiché quand une pulsation est détectée.

- Fra 18 -2019/06/03 14:04

MESURE DE LA TENSION

PressionsystoliquePression
diastolique
Indicateur depulsations irrégulières(IHB)

CH-650̲1901.indb 19

Français

4 Le résultat de la mesure est affiché.

Pulsation

Indicateur de mouvement du corps

L’indicateur de mouvement du corps est affiché surl’affichage des résultats de la mesure si vous avez bougé
votre main ou votre poignet pendant la mesure et qu’unevariation substantielle de pression a été détectée.
• Si l’indicateur de mouvement du corps est affiché
refaites la mesure de votre pression artérielle.
Si une pulsation irrégulière est détectée pendant lamesure, l’indicateur de pulsation irrégulière (IHB) est
affichée une fois que la mesure est terminée.
Un battement de cœur irrégulier est défini comme
un battement qui varie de ±25% à partir du milieu
d’un intervalle de 5 battements de cœur pendant une
mesure de la pression artérielle.
• La mesure peut ne pas être correct si les battements
de votre cœur varient de façon importante pendant
la mesure. Si l’indicateur de pulsation irrégulière(IHB) est affiché, refaites la mesure de votre pression
artérielle.

Attention

Si l’indicateur de pulsation irrégulière (IHB) apparaît
fréquemment, nous vous recommandons deconsulter un médecin pour examiner votre santé.

- Fra 19 -2019/06/03 14:04

MESURE DE LA TENSION

5 Arrêter la mesure.
• Appuyez sur la touche «START/STOP» pour éteindre l’appareil.

FrançaisCH-650̲1901.indb 20

FONCTION DE MISE HORS TENSION AUTOMATIQUE• Une fois la tension artérielle mesurée, appuyez sur la touche«START/STOP» pour éteindre l’appareil. Même si vous n’appuyez
pas sur la touche, la fonction d’extinction auto éteindra
automatiquement l’appareil au bout de 3 minutes.

RAPPEL DES DONNÉES MÉMORISÉES
60 résultats de mesures peuvent être mémorisés. La valeur moyenne
est calculée automatiquement pour vous aider à gérer votre santéquotidiennement.

COMMENT RAPPELER LES DONNÉES
1 Appuyez sur le bouton «MEMORY».

• «3A» qui indique la valeur moyenne apparaît dans la section gauchede l’affichage. Puis, la valeur moyenne basée sur les trois dernières
mesures est affichée. («3A» et la valeur moyenne n’apparaissent pas
s’il n’y a que deux mesures ou moins en mémoire.)- Fra 20 -2019/06/03 14:04

RAPPEL DES DONNÉES MÉMORISÉES

2 Chaque fois que vous appuyez sur le bouton, les données des mesures
Le numéro desdonnées est affichéavant les résultats
des mesures.

Françaissont rappelées dans l’ordre des plus récentes aux plus anciennes.

Les résultats des mesures sont affichés après le numéro des données.
• Les valeurs mémorisées sont numérotées dans l’ordre des mesures.
Par exemple quand 60 jeux de données sont mémorisés, les données
avec le numéro 1 représentent les données les plus récentes.
Numéro de données: 1, 2, 3, ... ... 59, 60ÇÇDonnées les plus récentes Données les plus anciennes

COMMENT EFFACER LES DONNÉES
1 Appuyez sur le bouton «MEMORY».
2 Tout en maintenant pressé le bouton «MEMORY», appuyez

sur le bouton «START/STOP» et maintenez-le enfoncé
pendant 3 secondes.
Toutes les données mémorisées sont effacées.
Maintenez
Un message
enfoncéindiquant que
pendant 3
toutes les données
secondes.
ont été effacéesde la mémoire
apparaît.
Effacé

CH-650̲1901.indb 21- Fra 21 -2019/06/03 14:04

INDICATIONS D’ERREURMARQUE AFFICHEE

FrançaisCH-650̲1901.indb 22

ETAT/CAUSEACTION CORRECTIVE

S’affiche lorsque lebrassard n’est pas
suffisamment serrée
autour du poignet.

Remettez le brassard
en place en la serrant.
Effectuez une nouvellemesure en veillant àdemeurer immobile.
(
Voir page
14 «FIXATION DU
BRASSARD».)

S’affiche si votre poignet
a bougé pendant lamesure; ou s’affichelorsque le brassard est
mal fixée.

Remettez le brassard
correctement en place.
Effectuez une nouvellemesure en veillant àdemeurer parfaitement
immobile. (Voirpage 14 «FIXATION DU
BRASSARD».)

S’affiche lorsque les pilessont déchargées.

Remplacez les deux
piles alcalines par despiles neuves. (Voirpage 13 «RETRAIT DESPILES».)

S’affiche lorsque lenombre de pulsations
tombe en dehors dela gamme de mesure(au-dessous de 39
battements/min. ou
au-dessus de 181
battements/min.)Remettez le brassard
correctement en place.
Respirez profondément afin
de vous détendre, et effectuez
une nouvelle mesure. (Voir page 16 «ASTUCESPOUR OBTENIR DES
MESURES DE PRECISION».)

L’appareil ne fonctionne
pas correctement.

Contactez le lieu d’achat
de l’appareil ou le centrede réparations le plus
proche. (Voir page
25 «AVANT DE SOLLICITER
UNE RÉPARATION OU DESTESTS».)- Fra 22 -2019/06/03 14:04

À PROPOS DE LA «TENSION ARTÉRIELLE»
Le cœur est une pompe qui assure la circulation du sang dans le corps. Le sang
est pompé à partir du cœur à pression constante en direction des artères. Cettepression est appelée tension artérielle et représente, en général, votre tension. La
tension est indiquée par divers types de pression comme : la pression systoliquequi se produit lorsque le cœur pompe le sang, et la pression diastolique qui se
produit lorsque le sang retourne vers le cœur.
(Systolique)(Diastolique)ArtèreAtriumgaucheVentricule
gaucheArtèreAtriumgaucheLes parois de l’artère
s’agrandissent.

Ventricule
gaucheFrançais

Qu’est-ce que la tension artérielle ?

Retour à l’état normal dela paroi de l’artère
Dilatation

Contraction

Votre tension change en permanence.

La tension varie en fonction de l’âge, du sexe et d’une variété d’autres facteurs. La
tension d’une personne est susceptible d’être influencée par le biorythme d’une
journée, par la position, par l’activité physique, par l’activité mentale, par le niveau
de stress et par la température de l’air. La tension artérielle d’une personne enbonne santé varie généralement au cours d’une journée.

[Exemple de variations de tension artérielle sur une journée]Pression artérielle
systolique

Valeur de tension (mmHg)

160
140
120
100
80
60

Pression artérielle
diastolique

40

Sommeil

12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11

Moment de la journée(Valeurs de la tension artérielle mesurées à des intervalles de cinq minutes sur une journée)l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329–44.

CH-650̲1901.indb 23- Fra 23 -2019/06/03 14:04

QUESTIONS ET RÉPONSES SUR LA TENSIONQ Pourquoi est-ce que la tension mesurée à la maison est différente decelle mesurée dans le cabinet d’un médecin ou à l’hôpital ?

A Votre tension est susceptible de varier en fonction de l’exercice, deFrançaisCH-650̲1901.indb 24

la température ambiante ou de votre état psychologique. Lorsqu’unmédecin ou une infirmière prennent votre tension, la valeur de tension
est probablement supérieure de 10 à 20 mmHg à la valeur habituelle
en raison de votre anxiété et/ou stress. Connaître ce qui influence votre
tension vous aidera à mieux surveiller votre santé.

Q Pourquoi la tension artérielle obtenue varie-t-elle à chaque mesure ?
A Notre tension artérielle est réglée par la fonction d’un nerf automatique.

La tension artérielle change à chaque battement de cœur. Nous pouvonspenser que notre tension artérielle est constante mais elle change si vousprenez plusieurs mesures. La tension artérielle est sensible au momentde la journée, comme par exemple le matin et l’après-midi, la saison et latempérature atmosphérique. De plus, la tension artérielle est facilement
influencée par le stress mental et les émotions. Elle a tendance à
augmenter lorsque vous vous sentez tendu et à baisser lorsque vousêtes détendu.

Q Quel est l’avantage de mesurer ma tension à domicile ?
A La tension mesurée à domicile offre des données fiables dans la mesureoù elle peut être réalisée dans un état stable et détendu. Les médecins
accordent une grande importance aux mesures de tension réalisées à
domicile. Vous devriez mesurer votre tension chaque jour à la mêmeheure et consigner les écarts de tension.

Consultez toujours votre médecin pour l’interprétation devos mesures de tension et pour déterminer le traitement
adapté.

- Fra 24 -2019/06/03 14:04

AVANT DE SOLLICITER UNE RÉPARATION OU DES TESTS
Vérifiez les éléments ci-dessous avant de faire tester ou réparer votre
appareil.

Aucune mesure
ne peut êtreprise.

Vérifications

RéponseVérifiez si les pilessont épuisées.

Remplacez toutes lespiles par des piles neuves.

Vérifiez si les pilessont installées
correctement(polarités etrespectées).

Installez les piles dans lebon sens.

Vérifiez si l’icôns’affiche.

Mettez le bracelet
correctement en place.

Français

Problème

L’affichage
ne s’allume
pas lorsque
vous appuyez
sur le bouton«START/STOP».

Vérifiez si le bracelet
est correctement
placé.
Etes-vous resté
calme pendant lamesure ?

Mesurez à nouveau votre
tension en veillant à rester
calme.

Remarquez qu’une mesure de tension peutêtre impossible pour quelqu’un dont le pouls
est extrêmement faible ou pour une personne
souffrant d’arythmie cardiaque (pulsations
irrégulières).
Vos mesures de tension sontanormalement élevées ou basses par
rapport à celles prises à l’hôpital.

Retirez les piles de l’unitéet remplacez les par desnouvelles.

Autres symptômesCH-650̲1901.indb 25

Voir page 24.

- Fra 25 -2019/06/03 14:04

SPÉCIFICATIONSFrançaisCH-650̲1901.indb 26

Numéro du modèle
Système de mesure
Affichage
Plage de circonférence du poignet
Pression
Plage demesurePouls
Pression
Précision
Pouls
Gonflement
Tension nominale
Échappement d’air
Alimentation
Durée des piles
Température
HumiditéConditions de Températurestockage
Humidité
Fonction mémoire
Protection contre les chocsélectriques
Niveau de protection
Mode de fonctionnement
Protection contre l’entrée d’eau
Dimensions
Poids
Durée de vie
Conditions
d’utilisation

CH-650Méthode oscillométrique
Affichage numérique
13,5 à 21,5 cm
0 à 280 mmHg
40 à 180 pulsation/min.
±3 mmHg±5% de la lecture
Gonflement automatique par pompe interne
CC 3V(: Courant direct)
Valve électromagnétique d’évacuation rapide
2 piles AAA (LR03)(180 mmHg, une
Env. 400 utilisations
fois par jour, 22°C)
10°C à 40°C
15% à 85% RH-20°C à 60°C
10% à 95%RH
Jusqu’à 60 mesures
Unité d’alimentation interne

Partie appliquée de type BF
Fonctionnement continu
IPX065(L) x 74(H) x 29(P) mm
Env. 103g piles exclure
5 ans
Ensemble composé de 2 piles AAA (LR03)
Accessoires
pour l’appareil, du mode d’emploi et d’unboîtier de rangement.
* Le brassard est la partie appliquée de ce dispositif.
* Plage de pression barométrique: 700 hPa à 1060 hPa*L a précision globale du système de ce dispositif es compatible avec
l’exigence de l’article 7.9 de EN1060-3.
EN1060-3 : Tensiomètres non invasifs Partie 3 (Norme européenne)- Fra 26 -2019/06/03 14:04

CH-650̲1901.indb 27

MEMO

2019/06/03 14:04

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 28

2019/06/03 14:04

CH-650̲1901.indb 1

中文

使用说明书
电子血压计

CH-650中国语

1901

2019/06/03 14:04

目录

中文
CH-650̲1901.indb 2

3 - 一般事项項5 - 使用及保养注意事项項
10 - 部件名称及布局
12 - 安装电池
13 - 更换电池
14 - 如何绑卷腕带
16 - 获得准确测量的提示
18 - 测量血压
20 - 调用存储器中的储存数据22 - 错误的显示图标23 - 关于“血压”
24 - 关于血压的一些问题和答案
25 - 要求修理或检测之前
26 - 规格

请确认以下物品均已齐全

血压计本体

收藏盒

（用于血压计）2枚AAA电池

使用说明书

-中2-

2019/06/03 14:04

一般事项
有关血压及血压测量的一般事项
迅速排放腕带中的气压时，请按“START/STOP”开关。
附送腕带适用于周长13.5厘米至21.5厘米的手腕。
绑卷腕带时，必须将血压计的本体放在手腕内侧。
本血压计之设计为使用于家庭使用，因此不宜作为诊断设备来用。
自我测量并非治疗！在任何情况下，均不可擅自更改医生处方的药物剂量。
6. 准备测量血压时，应该提前10至15分钟排尿，然后尽量保持安静。
7. 在测量前运动，进餐，饮酒，吸烟等，均可能影响测量结果。
8. 血压在一天中会不停变化，因此应该在每天同一时间测量。
9. 请勿太在意某次测量结果。坚持记录血压变化，多次测量结果才可说明问题。
10. 精神紧张往往会导致血压升高。

中文

1.
2.
3.
4.
5.

获得准确测量血压的方法
1. 在测量血压前，做五至六次深呼吸，然后放松。若在测量时情绪紧张，会造成测量结果不准确。
2. 如果你处于焦躁或恼怒状态，睡眠不足或便秘，或刚进行某些运动
或进餐后，血压均会升高。
3. 将腕带紧贴地绑卷在手腕上。（
参看第14页）4. 请勿在沐浴或饮酒后测量血压。
5. 应在测量血压前排尿。
6. 在20℃左右的室温下测量血压。请勿在室温过低（10℃以下）或过高（40℃以上）的情况下测量血压。
7. 请勿在饮用咖啡，茶等饮料或吸烟后立即测量血压。
8. 在放松，安静状态下测量血压。确保腕带中央与心脏齐高，请勿动手臂或讲话。
9. 请勿在卷上腕带数分钟后才测量血压，因为在这段时间内手腕被收紧，使血压升高，以致测量结果不准确。

CH-650̲1901.indb 3-中3-

2019/06/03 14:04

一般事项
10. 若显示出了“身体运动指示记号”或“不规则心跳（IHB）指示记
号”，那么请放松并重新测量您的血压。如果经常出现“心律不齐（IHB）指示记号”，那么建议您向医生咨询您的健康状况
参看第19页）。
（11. 下图表示按WHO标准划分的血压类型。
收缩压（mmHg）

中文
CH-650̲1901.indb 4

180

3级高血压（严重）2级高血压（一般）160
1级高血压（轻微）
140
130
120

备注：
本图表并不精确，但可作
为指引来了解非侵入式血压测量。本血压计仅供成
年人使用。

正常偏高
正常
理想
80 85 90

100

110

舒张压
（mmHg）

血压级别的定义及分类
12. 使用目的• 本设备是通过示波的方式进行测量的非侵入式血压计。本设备可自动测量收缩压（SYS）、舒张压（DIA）和脉搏。
• 本设备为在医生的指导下在家里检查人体健康情况的仪器，不能作为诊断仪器使用。
• 本设备不能用于婴儿或幼儿，而且无法进行自动循环测量。

每天于同一时间测量血压。
* 血压时常变化不定，故单独一次测量结果并无意义，长期搜集的数
据才具有价值。因此，应该坚持每天测量血压，尽量是每天于同一
时间测量血压。

-中4-2019/06/03 14:04

使用及保养注意事项
1. 如果你患有心脏病，高血压或其它循环系统疾病，请在使用血压计
前向医生咨询。
2. 在使用腕带时，如果感到腕带压力异常或发现任何异常情况，请立
即关闭电源或取出电池以减小压力，然后咨询购买血压计的销售中心。
3. 如认为测量的数值异常或测量令你感到不适，请停止使用并咨询医生。
4. 脉搏微弱或心律不齐的人士可能无法测量血压。
5. 反复测量血压后，某些人士可能会出现充血，肿胀等症状。
6. 频繁反复测量血压，则不能获得准确结果。应间隔1分钟左右。
7. 如果你的手臂患有严重血液循环病症，请在使用血压计前向医生咨
询，否则可能对你的健康造成危害。
8. 测量部位血液流量不够或心律极无规律的人士可能无法测量血压。
请向医生咨询，决定能否使用血压计。
9. 请勿将腕带卷在受伤的手腕上。
10. 请勿将腕带卷在正在接受点滴（静脉注射）或输血之类药物治疗的
手腕上，否则可能导致受伤或严重事故。
11. 请勿在麻醉气体等易燃气体附近使用血压计，以免产生火灾及爆炸危险。
12. 请勿在高氧室或氧气帐篷等氧气密集的环境中使用血压计，以免产生火灾及爆炸危险。
13. 请勿在血压计附近使用手机，否则可能导致故障。
14. 如果你使用心脏起搏器，请在使用血压计前向医生咨询。

CH-650̲1901.indb 5

中文

使用注意事项

-中5-

2019/06/03 14:04

使用及保养注意事项
15. 本血压计仅适用于测量血压，请勿用于其它用途。
16. 请务必使用经我们认证的零件和附件。零件和附件若末经认证可用
于本设备，则可能会使设备损坏。
17. 请勿在婴幼儿身上使用血压计。
18. 无法为患有心房或心室早期搏动或心房纤维性颤动病症的心律不齐
患者进行血压测量。

保养注意事项
中文
CH-650̲1901.indb 6

1. 请勿将血压计置于阳光直射，高温（60℃以上），低温（-20℃以下），高湿度（95%以上）或多尘的地方。
请一定要将血压计远离儿童、宠物或害虫存放。
2. 请勿跌落，撞击或震动血压计。
3. 如果长时间不使用血压计，请取出电池。
4. 请勿擅自拆卸本血压计。
5. 除非有需要，否则请勿弯曲腕带。
6. 如果血压计很脏，请用蘸有消毒酒精或中性去污剂的软布清洁，然后用干布擦净。
7. 切勿用稀释剂或苯擦洗血压计，否则会损坏血压计。
8. 请用拧干水的湿布擦拭腕带。用力摩擦可能会造成漏气。

-中6-

2019/06/03 14:04

使用及保养注意事项

警告请勿改装本装置。

符号说明CH-650̲1901.indb 7: BF型适用的部件: 保持干燥
: 警告: 使用前参 阅说明书: 设备符合 WEEE 指令
: 生产商

中文

: CE 标记表示向欧盟成员国的市场观察员提供信息。

本产品经EMC检测，符合EN60601-1-2标准。

警告这并不以任何方式保证本产品不会受到电磁干扰。请勿在高强度电磁环境下使用本产品。

-中7-

2019/06/03 14:04

使用及保养注意事项
正常保养和保养周期:本产品的使用期限较长，但通常建议每两年检查一次，以确保功能及性能完好。

保护自然环境：在弃置使用期限届满的电池或产品时，请遵守所在国家及／或当地回收法规，以协助保护自然环境。

WEEE 标志

中文
CH-650̲1901.indb 8

对本产品进行废弃处理时，请勿与普通家庭垃圾混在一起丢弃。根据WEEE 指令（指令 2012/19/EU）的规定，废旧电子产品有单独的回收系
统，仅在欧盟范围内有效。

非欧盟国家的处置须知如果您要丢弃废电池，请联系当地政府或经销商，询问正确的处置方法。

电池符号的说明
本符号可能与化学符号结合使用。这种情况下，处置电池必须符合所含化学物质
相关规定的要求。

欧洲参考标准：本血压计符合EN60601-1、EN1060-1、EN1060-3 和EN1060-4 标准。

校准：建议在通常情况下每两年检验一次本血压计，以确保正确的功能和测量精度。
详情请与销售商接洽。

-中8-

2019/06/03 14:04

使用及保养注意事项

名称：
地址：

CITIZEN SYSTEMS JAPAN CO.,LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan

工厂
CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen, Guangdong,
China

中文

名称：
地址：

欧洲代表：
名称：
地址：

EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague, The Netherlands• 有关技术数据，请联系我们的欧盟代表。

中国制造CH-650̲1901.indb 9-中9-

2019/06/03 14:04

部件名称及布局
部件名称
<本体>

显示屏

中文
CH-650̲1901.indb 10

存储器（MEMORY）
开关
启动/停止开关
待机

电池盒盖

腕带

- 中 10 -2019/06/03 14:04

部件名称及布局

<显示屏>

中文

收缩压

舒张压

心律不齐（IHB）
图标脉搏

身体移动指示符CH-650̲1901.indb 11

电池电量指示符电池电量不足时显示。
（
参看第13页）- 中 11 -2019/06/03 14:04

安装电池（使用血压计前，请先将随机附送的电池装入。）

1 打开电池盒盖。

中文CH-650̲1901.indb 12电池盒盖按下电池盒盖，沿箭头方向滑下盖子。

2 将电池置于电池盒内。
放入电池时，请注意正、负极符号* 首先放入电池的负极（ ）。

和。

首先放入靠近显示屏的那节电池。

- 中 12 -2019/06/03 14:04

安装电池

3 将电池盒的盖关上。

中文沿箭头方向滑入盖子。将盖子推
入至咔嗒到位。

更换电池• 当显示器上显示图标或无任何显示时，则须换上两枚新电池。
更换电池时，必须两枚电池一起更换。
• 请只使用碱性干电池。
• 请勿使用充电干电池。
• 请勿混合使用碱性干电池（LR03）和锰性干电池（R03）。
• 附送的电池仅供演示用途。电池寿命可能短于规定值。
• 丢弃废电池时，请遵守所在国家/地区的政府法规或环境部门的规
定。

CH-650̲1901.indb 13- 中 13 -2019/06/03 14:04

如何绑卷腕带
正确地将腕带绑卷在手腕上。如果腕带绑卷不正确，可能会导致测量不准确。

1 展开腕带。
2 将腕带绑卷在手腕上，血压计位于
中文
CH-650̲1901.indb 14

手腕内侧。
• 将腕带直接绑卷在裸露皮肤上。
• 腕带的设计适用于周长13.5至21.5厘
米的手腕。

3 拉动腕带末端，使腕带宽松地
贴在手腕上。

- 中 14 -2019/06/03 14:04

如何绑卷腕带

4 将腕带的顶缘调整到距离手腕皱褶1至

中文

2厘米左右的位置。

5 用魔术贴固定臂带。
• 如果腕带绑卷过松或过紧，可能会导致错误或读数不准确。

可用右腕来测量血压。
但是，左右手腕测得的血压可能会相差10mmHg，最好每次都用同一手
腕来测量血压。
为了获得准确测量结果• 腕带绑卷过松或过紧都可能会造成测量不准确。
• 测量时坐直。

CH-650̲1901.indb 15- 中 15 -2019/06/03 14:04

获得准确测量的提示
血压是随姿势、时间和许多其它因素的变动范围而变化的。最理想的
是每天同一时间按同一姿势测量血压。

➀ 确保姿势准确。
坐直，深呼吸，然后放松。

➁ 手掌朝上，手指放松。
中文CH-650̲1901.indb 16➂ 务必使腕带与您的心脏保
持在同一水平位置。

在手腕下垫上一块折叠的毛巾或类似物品。

- 中 16 -2019/06/03 14:04

获得准确测量的提示

为了获得准确测量结果

CH-650̲1901.indb 17

中文• 请勿紧握拳头。

• 请勿将腕带绑卷得过松。

• 请勿移动血压计的位置。

• 请勿在测量时移动身体或说话。

- 中 17 -2019/06/03 14:04

测量血压
1 将腕带绑卷在左腕上。
（

参看第14页“如何绑卷腕带”。）

2 确保姿势正确。
（

 参看第16页“获得准确测量的提示”。）

3 按下“START/STOP”键开始测量。
•

中文
CH-650̲1901.indb 18

所有数字显示均为初始显示，表示血压计工作正常。
同时会显示标记，但并不代表
需要更换电池。

检测到脉搏时，显示。

- 中 18 -2019/06/03 14:04

测量血压

4 测量结果显示。
收缩压

中文

舒张压
心律不齐（IHB）
图标

CH-650̲1901.indb 19脉搏身体移动指示符如果测量期间移动了手或手腕，则会使测量到的血压随着发生波动，测量结果画面中会显示身体移动图标。
• 如果显示身体移动图标，请重新测量血压。
如果测量期间检测到心律不齐，在测量完成后会显
示心律不齐（IHB）图标。
非正常心跳指在血压测量过程中出现的与中间连续
5次心跳间隔变化量在±25%的心跳。
• 如果测量期间心率波动很大，则测量结果可能会
不准确。如果显示心律不齐（IHB）图标，请重新测量血压。

注意

如果频繁出现心律不齐（IHB）图标，建议您向医生咨询您的健康状况。

- 中 19 -2019/06/03 14:04

测量血压

5 结束量度。
• 按着“START/STOP”键以关闭血压计。

自动关机功能• 在量度了血压之后，按“START/STOP”键就会关闭血压计。但是，如果不按该键，血压计的“自动关机”（Auto OFF）功能会在3分钟
之后自动关闭血压计。

中文
CH-650̲1901.indb 20调用存储器中的储存数据存储器可以储存60次测量结果。血压计会自动计算平均值，帮助您管理日常健康。

如何调用数据1 按“MEMORY”重读开关。

• “3A”表示平均值，显示在显示屏的左侧部分。然后，显示最近
三次测量的平均值。（如果存储器中仅储存了两次或更少的测量数据，将不显示“3A”和平均值。）- 中 20 -2019/06/03 14:04

调用存储器中的储存数据2 每次按此开关，血压计会按照最新到最早的次序显示测量数据。

中文测量结果前显示数
据编号。

测量结果显示在数据编号后。
• 记忆体内的数值是按量度次序而储存的，例如，当记忆体储存了60
组数据，最新一组数据的数据编号就是1。
数据编号：1, 2, 3, ... ... 59, 60
Ç
Ç
最新的数据
最早的数据

如何删除数据1 按“MEMORY”开关。
2 按“MEMORY”开关的同时，按住“START/STOP”键3秒钟以上。
所有储存的数据即被删除。

CH-650̲1901.indb 21

按住3秒钟

删除完毕

显示一条消息，告知已删除存储
器中的所有数据。

- 中 21 -2019/06/03 14:04

错误的显示图标
显示标记

中文
CH-650̲1901.indb 22

状况/原因

纠正措施

腕带绑卷过松时显示。

重新紧贴地绑卷腕带。重
新测量读数，确保您保持
不动。（
参看第14
页“如何绑卷腕带”。）

测量期间手腕移动时显
重新正确绑卷腕带。重示；或者腕带绑卷不正确 新测量读数，确保您完时显示。
全保持不动。（
参看第14页“如何绑卷腕带”。）

电池用完时显示。

更新两节新的碱性电池。
（
参看第13页“更换电池”。）

脉率超出测量范围（低于39次/分钟或高于
181次/分钟）时显示。

重新正确绑卷腕带。深呼吸放松，再重新测量。
（
参看第16页“获得准确测量的提示”。）

仪器无法正常工作。

咨询购买血压计的商店或当地服务中心（参看第25页“要求修理或检测之前”。）- 中 22 -2019/06/03 14:04

关于“血压”
何谓血压？
心脏像一个泵，推动血液在身体内流动。心脏将血液泵到动脉，这种压力称为动脉血压，一般来说，代表血压。血压以数种压力显示，如：心脏收缩压，当心脏泵血时出现；而心脏舒张压，当血液回流心脏时出现。
(心脏收缩压)(心脏舒张压)
动脉

左心房

动脉

左心室

中文

左心房

左心室
动脉壁扩张。

动脉壁恢复正常状态。

扩张扩张

血压是会改变的
血压是随年龄、性别和许多其它因素的变动范围而变化的。血压很容易受一天的生物节律以及姿势、身体活动、心理活动、紧张程度和气温影响。一般认为正常
人一天的血压变化量在。

[一天内血压波动的图例]
心脏收缩压

160

血压值(mmHg)

140
120
100
80
60

心脏舒张压

40

睡眠
12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11

一天之中的时间

（一天内每隔五分钟测得的血压值）l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329-44.

CH-650̲1901.indb 23- 中 23 -2019/06/03 14:04

关于血压的一些问题和答案
Q 为何在家中测量的血压有别于医生或医院测量的？
A 你的血压可能是随运动、环境温度或心理状态而变化的。当医生或护士为你测量血压时，忧虑和紧张的心情很容易使血压数值较平常高出10至
20mmHg。了解影响血压的因素将更有利于你保护健康。

Q 为何每次量度的血压都有分别？
A 我们的血压受自动神经功能所调节。心脏每次跳动的血压都会不同。我中文
CH-650̲1901.indb 24

们都很容易以为血压是持续不变的，但重复量度时却会发现当中存在差别。血压很容易受一天之中的时间，例如早上和午间、季节和大气温度所影响。此外，血压也很容易受心理压力或情绪起伏影响。紧张的时候
血压通常会上升，放松时通常会下降。

Q 在家测量血压的益处？
A 在家测量的血压较为可靠，因为测量时的环境较为稳定和轻松。医生很重视在家测量血压的数据，建议每天于相同时间测量血压，并记录血压的变化。

请经常向医生咨询，请医生分析你的血压值和提供适当的治疗方法。

- 中 24 -2019/06/03 14:04

要求修理或检测之前
存在问题

各种检查

相应措施

按“START/
STOP”开关时,显示屏出现空白显示。

检查电池电量是否
耗尽。

将所有电池更换为新电池。

检查电池是否未以正
）
确的极性（
安装。

以正确的极性装入电池。

标记是否
检查
显示。

正确地绑卷腕带。

不能测量血压。

中文

将血压计送往修理／检查之前请先检查下列各项。

检查腕带的绑卷是
否恰当。
测量时你的心情是否平静？

再次测量，务必保持安静。

注意：如果脉搏过于微弱或心律不齐（脉搏不规则），血压计或许不能测量其血压。
血压数值异常高于或异常低于医院的
测量数值。
其它情况

CH-650̲1901.indb 25

参看第24页。
取出仪器内的电池，更换新电池。

- 中 25 -2019/06/03 14:04

规格
型号
测量系统
显示
腕围范围
测量范围
准确度

中文
CH-650̲1901.indb 26

血压
脉搏
血压
脉搏

充气功能
额定电压
排气
电源
电池寿命
工作条件
存放条件
记忆功能
触电保护
防护等级
操作模式
进水防护
体积
重量
使用寿命
附件

温度
湿度
温度
湿度

CH-650示波法
数字显示
13.5至21.5厘米0至280毫米汞柱40 - 180次/分±3毫米汞柱读数的±5%
由内部充气泵自动充气
DC 3V（: 直流电）
快速排气电磁阀
2节AAA电池（LR03）（180mmHg,
约400次
1次/天, 22°C）
10至40°C
15至85% RH-20至60°C
10至95% RH
最多60次测量结果内部电源设备
BF 型应用部分
连续操作
IPX065（宽）× 74（长）× 29（高）毫米
约103克（没有电池）5年1套  血压计用2节AAA电池（LR03）、使用说明书、收藏盒* 本设备随附的部件为臂带。
* 气压的范围为700 hPa 至 1060 hPa* 本设备整个系统的精确率符合EN1060-3中第7.9条的要求。
EN1060-3：非侵入式血压计 第3部分（欧盟标准）- 中 26 -2019/06/03 14:04

CH-650̲1901.indb 27

MEMO

2019/06/03 14:04

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 28

2019/06/03 14:04

ИНСТРУКЦИЯ ПО ЭКСПЛУАТАЦИИ
Прибор для измерения артериального
давления и частоты пульса цифровой

CH-650̲1901.indb 1

Pyсский

CH-650Pyсский

1901

2019/06/03 14:04

Содержание
3 - ОБЩИЕ СВЕДЕНИЯ
5 - МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ ИОБСЛУЖИВАНИИ
10 - НАЗВАНИЯ И РАСПОЛОЖЕНИЕ ЧАСТЕЙ
12 - УСТАНОВКА БАТАРЕЕК
13 - ИЗВЛЕЧЕНИЕ БАТАРЕЕК
14 - НАДЕВАНИЕ МАНЖЕТЫ НА ЗАПЯСТЬЕ
16 - СОВЕТЫ ПО ПРАВИЛЬНОМУ ВЫПОЛНЕНИЮ ЗАМЕРОВ
18 - ИЗМЕРЕНИЕ АРТЕРИАЛЬНОГО ДАВЛЕНИЯ

Pyсский
CH-650̲1901.indb 2

20 - ВЫЗОВ ДАННЫХ, СОХРАНЕННЫХ В ПАМЯТИ
22 - ИНДИКАЦИЯ ОШИБОК
23 - АРТЕРИАЛЬНОЕ ДАВЛЕНИЕ
24 - ВОПРОСЫ И ОТВЕТЫ ОТНОСИТЕЛЬНО
АРТЕРИАЛЬНОГО ДАВЛЕНИЯ
25 - ПРЕЖДЕ ЧЕМ ОБРАЗАТЬСЯ ЗА РЕМОНТОМ ИЛИПРОВЕРКОЙ
26 - ТЕХНИЧЕСКИЕ ХАРАКТЕРИСТИКИ
Убедитесь в наличии всех компонентов, перечисленных нижеАппарат
монитора
кровяного
давления

Футляр дляхранения

(для прибора)2 батареи
размера AAAИнструкция поэксплуатации

- Рус 2 -2019/06/03 14:04

ОБЩИЕ СВЕДЕНИЯ
1. Чтобы быстро выпустить воздух из манжеты, нажмите кнопку
“START/STOP”.
2. Прикрепленная манжета может использоваться для запястья собхватом 13,5-21,5 см.
3. Во время закрепления манжеты корпус прибора для измерения
артериального давления должен располагаться на внутренней
стороне запястья.
4. Данный прибор предназначен не для использования в качестве
диагностического устройства, а для домашнего применения.
5. Самостоятельное измерение давления не заменяет лечения!
Изменение дозировки лекарств следует производить только поуказанию врача.
6. Перед измерением артериального давления следует опорожнить
мочевой пузырь и спокойно посидеть 10-15 минут.
7. Физические упражнения, прием пищи, напитков, курение и другие
факторы могут отразиться на результатах измерения.
8. В течение дня артериальное давление постоянно меняется. Измерения
следует производить ежедневно в одно и то же время.
9. НЕ преувеличивайте значение результатов единичного измерения.
Записывайте изменения артериального давления. Полную картину может
дать только большое количество собранных данных.
10. Эмоциональное потрясение может вызвать повышение
артериального давления.

Pyсский

Общие сведения об артериальном давлении и его измеренииКак правильно измерять артериальное давление

1. Перед измерением давления сделайте 5-6 глубоких вдохов ивыдохов и расслабьтесь. Если вы напряжены во время измерения,результат будет неправильным.
2. Артериальное давление может подняться, если вы чем-тообеспокоены, раздражены, не выспались, страдаете запором, а также
после приема пищи и выполнения физических упражнений.
3. Плотно оберните манжету вокруг запястья. (См. стр. 14)
4. НЕ следует измерять артериальное давление после приема ванныи употребления спиртных напитков.
5. Если вы ощущаете потребность сходить в туалет, это следует
сделать до измерения давления.
6. Измерять давление рекомендуется в помещении при температуре
около 20°C. НЕ измеряйте давление при слишком низкой (ниже 10°C) ивысокой (выше 40°C) температуре в помещении.
7. НЕ измеряйте артериальное давление после употребления кофе,чая или курения.

CH-650̲1901.indb 3- Рус 3 -2019/06/03 14:04

ОБЩИЕ СВЕДЕНИЯ

Pyсский
CH-650̲1901.indb 4

8. Измеряйте артериальное давление в расслабленном, спокойном
состоянии. Центр манжеты должен находиться на уровне сердца.
НЕ двигайте рукой и НЕ разговаривайте.
9. НЕ следует измерять артериальное давление спустя несколько
минут после надевания манжеты, так как давление в рукеповысится, и результаты измерения будут неправильными.
10. В случае отображения индикатора движения тела или индикатора
нерегулярного сердцебиения (IHB), расслабьтесь и проведите
повторное измерение артериального давления. В случае
частого появления индикатора нерегулярного сердцебиения
(IHB), рекомендуется проконсультироваться с врачом по поводу
См. стр. 19)
состояния здоровья. (11. На следующем рисунке показана классификация артериального
давления согласно стандартам ВОЗ.
СИСТОЛИЧЕСКОЕ (мм рт. ст.)180Гипертония, степень 3 (тяжелая)Гипертония, степень 2 (умеренная)

160
140
130
120

Гипертония, степень 1(слабая)Примечания:График не является точным,но может испол татов
неинвазивного измерения
артериального давления.
Прибор предназначен длявзрослых пользователей.

Повышенное
Нормальное
Оптимальное

80 85 90100
110 ДИАСТОЛИЧЕСКОЕ (мм рт. ст.)Определение и классификация уровней артериального давления
12. Предполагаемое использование• Данное устройство предназначено для неинвазивного измерения
кровяного давления осциллометрическим способом. Устройство
может автоматически измерять систолическое давление крови(SYS), диастолическое давление крови (DIA) и частоту пульса.
• Устройство предназначено для проверки состояния здоровьяв домашних условиях и не является диагностическим
приспособлением.
• Устройство не предназначено для использования нановорожденных или младенцах и не может использоваться дляавтоматического измерения циклов.

- Рус 4 -2019/06/03 14:04

ОБЩИЕ СВЕДЕНИЯ

Рекомендуется регулярно измерять артериальное
давление в одно и то же время.

* Артериальное давление постоянно меняется. Это означает,что данные, собранные за продолжительный период времени,имеют большую значимость, чем единичное измерение.
Поэтому следует измерять артериальное давление ежедневно.
В идеале, рекомендуется измерять артериальное давление водно и то же время каждый день.

МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ И ОБСЛУЖИВАНИИ
1. Если вы страдаете заболеваниями сердца, гипертониейили другими сосудистыми заболеваниями, сначала
проконсультируйтесь со своим врачом.
2. Если вы почувствуете анормальное давление в манжетеили какие-либо другие ненормальности в работе во время
использования манжеты, немедленно уменьшите давление
выключите прибор, выньте батарейки и обратитесь законсультацией в место покупки прибора.
3. Если вы полагаете, что полученные значения измерений
являются ненормальными или при измерении вы чувствуете
себя нехорошо, обратитесь к своему врачу.
4. Измерение артериального давления у людей со слабым
пульсом или аритмией может оказаться невозможным.
5. Многократное измерение давления у некоторых людей
может привести к таким проблемам, как закупорка сосудов иопухание.
6. При повторных часто выполняемых измерениях
артериального давления могут быть получены неправильные
результаты. Перед повторным измерением следует подождатьне менее1 минуты.
7. Проконсультируйтесь со своим врачом перед использованием
прибора, если вы страдаете серьезными нарушениями
кровообращения в области рук. Невыполнение этого
требования может представлять опасность для здоровья.
8. Измерение артериального давления при недостаточном
кровотоке в области выполнения измерения или причастом либо нерегулярном сердцебиении может оказаться
невозможным. Проконсультируйтесь со своим врачом.

CH-650̲1901.indb 5

Pyсский

Меры предосторожности при использовании

- Рус 5 -2019/06/03 14:04

МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ И ОБСЛУЖИВАНИИ

Pyсский
CH-650̲1901.indb 6

9. НЕ надевайте манжету на пораненное запястье.
10. НЕ надевайте манжету на запястье под капельницей(во время выполнения внутривенной инфузии) или припереливании крови. Несоблюдение этого правила может
привести к травме или другим тяжелым последствиям.
11. НЕ пользуйтесь прибором в помещениях с горючими газами,такими как анестезирующие газы, поскольку существует
опасность возгорания и взрыва.
12. НЕ пользуйтесь прибором в помещениях с повышенным
содержанием кислорода, таких как барокамера иликислородная палатка, поскольку существует опасность
возгорания.
13. НЕ пользуйтесь мобильным телефоном около прибора. Этоможет привести к неправильной работе прибора.
14. Лицам, использующим кардиостимуляторы, следует
предварительно проконсультироваться с врачом.
15. Этот прибор должен использоваться только для измерения
артериального давления. НЕ используйте его для других целей.
16. Обязательно используйте только разрешенные детали ипринадлежности. Детали и принадлежности, не одобренныедля использования с данным устройством, могут привести кповреждению прибора.
17. НЕ используйте данный прибор для измерения давления умладенцев.
18. Измерение артериального давления может бытьневозможным для людей с общими аритмиями, такимикак артериальная или желудочковая экстрасистолия илимерцательная аритмия.

Меры предосторожности при обслуживании

1. Прибор для измерения артериального давления следует
оберегать от воздействия прямого солнечного света, высокой
температуры (выше 60°C), низкой температуры (ниже -20°C),высокой относительной влажности (свыше 95%) и пыли.
Храните устройство для измерения кровяного давления внедоступном для детей, животных и насекомых месте.
2. Оберегайте прибор от вибрации и других ударов и толчков, нероняйте его.
3. Выньте батарейки, если прибор не используется в течение
продолжительного периода времени.
4. НЕ разбирайте прибор.

- Рус 6 -2019/06/03 14:04

МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ И ОБСЛУЖИВАНИИ
5. НЕ сгибайте манжету слишком сильно.
6. При сильном загрязнении прибора протрите его салфеткой,смоченной медицинским спиртом или нейтральным моющим
средством. Затем насухо вытрите прибор сухой материей.
7. НИКОГДА не используйте для очистки прибора разбавительили бензин, поскольку эти вещества могут повредить его.
8. Для очистки манжеты протирайте ее влажной тканью. Нетрите манжету сильно, поскольку это может стать причиной
утечки воздуха.

Предупреждение

Модификации данного устройства недопустимы.

CH-650̲1901.indb 7: Маркировка CE свидетельствует о соответствии
устройства основным требованиям директив ЕС.
: Рабочая часть типа BF: Хранить в сухом месте
: Внимание
: Перед использованием сверьтесь с руководством
: Устройство соответствует директиве WEEE
: Производитель

Предупреждение

Pyсский

Объяснение символов

Устройство было протестировано иратифицировано в соответствии состандартом EN60601-1-2 для ЭМС.
Это никоим образом не гарантирует
неподверженность устройства
электромагнитным помехам. Избегайте
использования устройства в среде
электромагнитного излучения.

- Рус 7 -2019/06/03 14:04

МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ И ОБСЛУЖИВАНИИ
Характер и частота обслуживания:Этот прибор предназначен для длительного использования, однако,рекомендуется осуществлять проверку его работоспособности каждые двагода.

Защита окружающей среды:Оказывайте содействие в защите окружающей среды, соблюдая
государственные законы и/или местные правила при утилизации батарейкии изделия по окончании их срока службы.

ЗНАК WEEE

Pyсский
CH-650̲1901.indb 8В случае утилизации данного прибора не выбрасывайте еговместе с обычными бытовыми отходами. Существует система
раздельного сбора использованных электронных изделий всоответствии с законодательством по реализации ДирективыWEEE (Директива 2012/19/EU), которая имеет силу только вЕвропейском Союзе.

Информация относительно утилизации в других странах запределами Европейского Союза.

Если вам необходимо утилизировать использованные батареи, обратитесьв местные органы власти или к продавцу, и спросите о правильном способе
утилизации.

Примечание относительно символа батареи.

Данный символ может использоваться в сочетании с символом химического
вещества. В этом случае к нему применяется требование, предусмотренное
директивой относительно данного химического вещества.

Европейский стандарт:Данный аппарат для измерения кровяного давления соответствует
стандарту EN60601-1, EN1060-1, EN1060-3, EN1060-4.

Градуировка:Монитор аппарата для измерения кровяного давления рекомендуется
проверять каждые два года, чтобы быть уверенным в правильности егоработы и точности показаний. Обратитесь к дистрибьютору.

- Рус 8 -2019/06/03 14:04

МЕРЫ ПРЕДОСТОРОЖНОСТИ ПРИ ИСПОЛЬЗОВАНИИ И ОБСЛУЖИВАНИИ

Наименование : CITIZEN SYSTEMS JAPAN CO., LTD.
Адрес :
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, JapanЗаводНаименование : CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Адрес :
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen,
Guangdong, ChinaНаименование : EMERGO EUROPE
Адрес :
Prinsessegracht 20, 2514 AP The Hague,
The Netherlands• Для получения информации о технических
данных обратитесь к европейскому
представителю компании.

Сделано в Китае

CH-650̲1901.indb 9

Pyсский

Представительство в Европе:- Рус 9 -2019/06/03 14:04

НАЗВАНИЯ И РАСПОЛОЖЕНИЕ ЧАСТЕЙ
Названия частей
<Основной аппарат>Дисплей

Pyсский
CH-650̲1901.indb 10Кнопка MEMORYКнопка
START/STOPРежим готовности

Крышка
батарейного
отсека

Манжета на запястье

- Рус 10 -2019/06/03 14:04

НАЗВАНИЯ И РАСПОЛОЖЕНИЕ ЧАСТЕЙ

<Дисплей>Диастолическое
артериальное
давление

Pyсский

Систолическое
артериальное
давление

Индикатор
нерегулярного
сердцебиения (IHB)
ПульсИндикатор движения телаCH-650̲1901.indb 11Индикатор уровня заряда батареи
Появляется в случае разрядки
батарей.
См. стр. 13)(- Рус 11 -2019/06/03 14:04

УСТАНОВКА БАТАРЕЕК
(Перед использованием сфигмоманометра необходимо установитьв батарейный отсек батарейки, поставляемые с прибором).

1 Откройте крышку батарейного отсека.

Pyсский
CH-650̲1901.indb 12Крышка
батарейного
отсека

Нажав на крышку
батарейного отсека,сдвиньте ее внаправлении стрелки.

2 Установите батарейки в отсек.

Установите батарейки в отсек, обратите внимание наполярность: символы и .
* Батареи вставляются сначала отрицательной стороной

.

Сначала вставьте батарею,расположенную ближе к дисплею.

- Рус 12 -2019/06/03 14:04

УСТАНОВКА БАТАРЕЕК

3 Закройте крышку батарейного отсека.

ИЗВЛЕЧЕНИЕ БАТАРЕЕК
• При отображении символа
, или в случае отсутствия
изображений на дисплее, замените старые батареи двумя.
новыми. Все четыре батареи следует заменять одновременно.
• Используйте только щелочные батареи.
• Не используйте перезаряжаемые батарейки.
• Не используйте одновременно сухие щелочные (LR03) имарганцевые батарейки (R03).
• Прилагаемые батареи предназначены только для демонстрационных
целей. Срок службы батарей может быть короче указанного.
• При утилизации использованных батарей соблюдайте государственные
нормы или стандарты общественных институтов по охране окружающей
среды, действующие в вашей стране/вашем регионе.

CH-650̲1901.indb 13

Pyсский

Сдвиньте крышку в направлении
стрелки. Надавите на крышку дощелчка.

- Рус 13 -2019/06/03 14:04

НАДЕВАНИЕ МАНЖЕТЫ НА ЗАПЯСТЬЕ
Оберните манжету надлежащим образом вокруг запястья.
Ненадлежащее обертывание манжеты может привести кошибкам измерения.

1 Откройте манжету для надеванияна запястье.

2 Оберните манжету вокруг левого
запястья таким образом, чтобы
прибор был расположен свнутренней стороны запястья.

Pyсский
CH-650̲1901.indb 14• Надевайте манжету на открытый
участок кожи запястья.
• Манжета для надевания назапястье рассчитана на запястьяс обхватом 13,5 - 21,5 см.

3 Свободно отрегулируйте
манжету на запястье,потянув конец манжеты.

- Рус 14 -2019/06/03 14:04

НАДЕВАНИЕ МАНЖЕТЫ НА ЗАПЯСТЬЕ

4 Расположите верхний крайманжеты приблизительно в 1 - 2
см от сгиба запястья.

5 Зафиксируйте манжету назапястье с помощью
застежки-липучки.

Pyсский

1 - 2 см• Если манжета на запястье
будет обернута слишком
свободно или слишком
туго, это может привестик ошибкам или неверным
результатам.

Вы можете измерять артериальное давление на правом
запястье.
Однако артериальное давление на левой и правой руке может
отличаться примерно на 10 мм рт. ст., поэтому лучше измерять
артериальное давление каждый раз на одном и том же запястье.
Для получения точных измерений
• Слишком свободное или слишком тугое обертывание манжеты
может помешать правильному измерению.
• Сидите прямо во время измерения.

CH-650̲1901.indb 15- Рус 15 -2019/06/03 14:04

СОВЕТЫ ПО ПРАВИЛЬНОМУ ВЫПОЛНЕНИЮ ЗАМЕРОВ
Артериальное давление зависит от положения, времени суток имножества других факторов. В идеале, рекомендуется измерять
артериальное давление в одном и том же положении в одно и тоже время каждый день.

➀ Проверьте правильность позы.

Сядьте прямо, сделайте глубокий вдох ирасслабьтесь.

➁ Расположите руку ладонью

вверх и расслабьте пальцы.

Pyсский
CH-650̲1901.indb 16➂ Важно, чтобы манжета
находилась на одном
уровне с сердцем.

Поместите под рукусложенное полотенце илианалогичный предмет.

- Рус 16 -2019/06/03 14:04

СОВЕТЫ ПО ПРАВИЛЬНОМУ ВЫПОЛНЕНИЮ ЗАМЕРОВДля получения правильных измерений

CH-650̲1901.indb 17• Не обертывайте манжету назапястье слишком свободно.

Pyсский

• Не сжимайте крепко кулак.

• Не меняйте положения прибора.

• Не двигайтесь и не разговаривайтево время измерения

- Рус 17 -2019/06/03 14:04

ИЗМЕРЕНИЕ АРТЕРИАЛЬНОГО ДАВЛЕНИЯ
1 Оберните манжету вокруг левого запястья.
(См. стр. 14 “НАДЕВАНИЕ МАНЖЕТЫ НА ЗАПЯСТЬЕ”.)

2 Проверьте правильность позы.
( м. стр. 16 “СОВЕТЫ ПО ПРАВИЛЬНОМУ ВЫПОЛНЕНИЮ
СЗАМЕРОВ”.)

3 Нажмите кнопку “START/STOP”, чтобы

Pyсский
CH-650̲1901.indb 18

начать измерение.
• Все символы отображаются в качестве
начальной индикации, показывая, чтоприбор работает нормально.
Появится также знак, однако этоне означает, что требуется замена
батареи.

отображается при обнаружении пульса.

- Рус 18 -2019/06/03 14:04

ИЗМЕРЕНИЕ АРТЕРИАЛЬНОГО ДАВЛЕНИЯ

4 Будут отображены результаты измерений.
Систолическое
артериальное
давление

Индикатор
нерегулярного
сердцебиения (IHB)

CH-650̲1901.indb 19ПульсИндикатор движения телаИндикатор движения тела будет отображен на дисплее
результатов измерения, если вы переместите руку илизапястье во время измерения, что приведет к обнаружению
значительного изменения давлени.
• В случае отображения индикатора движения тела,проведите повторное измерение артериального давления.

Pyсский

Диастолическое
артериальное
давлениеВ случае обнаружения нерегулярного сердцебиения вовремя измерения, после завершения измерения будет
отображен индикатор нерегулярного сердцебиения (IHB).
Нерегулярным считается сердцебиение, отличающееся
на ±25% от 5 средних ударов сердца на протяжении
измерения артериального давления.
• В случае значительных колебаний сердцебиенияво время измерения, его результаты могут бытьневерными. В случае отображения индикатора
нерегулярного сердцебиения (IHB), проведите
повторное измерение артериального давления.

ОсторожноВ случае частого появления индикатора
нерегулярного сердцебиения (IHB),рекомендуется проконсультироваться сврачом по поводу состояния здоровья.

- Рус 19 -2019/06/03 14:04

ИЗМЕРЕНИЕ АРТЕРИАЛЬНОГО ДАВЛЕНИЯ

5 Завершение измерения.
• Чтобы отключить монитор, нажмите кнопку “START/STOP”.

АВТОМАТИЧЕСКОЕ ИЗМЕНЕНИЕ РЕЖИМА

• После выполнения измерения кровяного давления нажатие
переключателя “START/STOP” отключит монитор. Однако
если Вы не нажмете данный переключатель, функция
автоматического отключения монитора (Auto OFF)автоматически отключит монитор по прошествии 3 минут.

Pyсский
CH-650̲1901.indb 20ВЫЗОВ ДАННЫХ, СОХРАНЕННЫХ В ПАМЯТИВ памяти можно сохранить 60 результатов измерения.
Среднее значение вычисляется автоматически, что помогает
ежедневно контролировать состояние здоровья.

ВЫЗОВ ДАННЫХ

1 Нажмите кнопку вызова данных из памяти “MEMORY”.

• В левой части дисплея появится индикация “3A”, обозначающая среднее
значение. Затем будет отображено среднее значение на основе последних
трех измерений. (Индикация “3A” и среднее значение не появятся, если впамяти сохранено два или меньшее число измерений.)- Рус 20 -2019/06/03 14:04

ВЫЗОВ ДАННЫХ, СОХРАНЕННЫХ В ПАМЯТИ

2 При каждом нажатии переключателя будут отображаться данные
измерений в порядке от наиболее новых к наиболее старым.

Результаты измерений будут отображены после номера данных.
• Значения, сохраненные в памяти, нумеруются в порядке проведения
измерений. Например, в случае сохранения в памяти 60 наборов
данных, номер данных 1 будет представлять самые последние данные.
Номер данных: 1, 2, 3, ...
... 59, 60ÇÇСамые последние Более ранние
данные
данные

Pyсский

Перед результатами
измерений будет
отображаться
номер данных.

УДАЛЕНИЕ ДАННЫХ

1 Нажмите кнопку вызова данных из памяти “MEMORY”.
2 Во время нажатия переключателя “MEMORY”, нажмите и удерживайте
переключатель “START/STOP” в течение более чем 3 секунд.
Все сохраненные данные будут удалены.

CH-650̲1901.indb 21Нажмите иудерживайтев течение 3секунд

Удалено

Будет отображено
сообщение,информирующеео том, что вседанные былиудалены изпамяти.

- Рус 21 -2019/06/03 14:04

ИНДИКАЦИЯ ОШИБОК
ИНДИКАЦИЯ НА ДИСПЛЕЕ

Pyсский
CH-650̲1901.indb 22СОСТОЯНИЕ/ПРИЧИНА

МЕТОД УСТРАНЕНИЯ

Появляется в томслучае, если манжетана запястье обернута
слишком свободно.

Повторно плотно
оберните манжету назапястье. Выполните ещеодно измерение, сохраняя
неподвижное положение.
См. стр.14(“НАДЕВАНИЕ МАНЖЕТЫНА ЗАПЯСТЬЕ” )Появляется в случае
перемещения запястьяво время измерения.;илиПоявляется в случае,если манжета обернута
неправильно.

Повторно правильно оберните
манжету на запястье.
Выполните еще одноизмерение, сохраняя полностью
неподвижное положение.
См. стр.14(“НАДЕВАНИЕ МАНЖЕТЫ НАЗАПЯСТЬЕ” )Появляется в случае
разрядки батарей.

Замените двещелочные батареи нановые.
См. стр.13(“ИЗВЛЕЧЕНИЕ
БАТАРЕЕК”.)Появляется в случае,если частота пульса
выходит за пределы
диапазона измерений
(ниже 39 ударов/мин. иливыше 181 удара/мин.)Повторно надлежащим
образом оберните манжету назапястье. Сделайте глубокий
вдох для расслабления иповторите измерение.
См. стр.16 “СОВЕТЫ(ПО ПРАВИЛЬНОМУ
ВЫПОЛНЕНИЮ ЗАМЕРОВ”.)Прибор нефункционирует
надлежащим образом.

Обратитесь в магазин, гдебыл приобретен прибор, илив местный сервисный центр.
См. стр.25 “ПРЕЖДЕ(ЧЕМ ОБРАЗАТЬСЯЗА РЕМОНТОМ ИЛИПРОВЕРКОЙ”.)- Рус 22 -2019/06/03 14:04

АРТЕРИАЛЬНОЕ ДАВЛЕНИЕЧто такое артериальное давление?
Сердце – это насос, который подает кровь к различным частям тела. Кровь при постоянном давлении
прокачивается через сердце в артерии. Это давление называется артериальным и отражает, в общем,давление крови. Артериальное давление характеризуется несколькими параметрами, такими как:систолическое давление, возникающее, когда сердце качает кровь, и диастолическое давление,возникающее, когда кровь возвращается в кровь.
(Диастолическое)(Систолическое)Артерия

Артерия
Левоепредсердие

Стенка артерии
расширяется.

Левыйжелудочек

Сжатие

Стенка артерии возвращаетсяв нормальное состояние.
Расширение

Артериальное давление постоянно изменяется.

Pyсский

Левыйжелудочек

Левоепредсердие

Артериальное давление зависит от возраста, пола и ряда других факторов. На негооказывают влияние дневные биоритмы, положение тела, физическая деятельность,психическая активность, уровень стресса и даже температура воздуха. В течениедня артериальное давление здорового человека обычно может изменяться.

Значение артериального давления (mmHg)[Пример изменений артериального давления в течение дня]Систолическое
артериальное давление

160
140
120
100
80
60
40

Во сне

Диастолическое
артериальное давление

12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11Время суток

(Артериальное давление, измеренное с пятиминутным интервалом втечение всего дня)CH-650̲1901.indb 23

l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329–44.
.
- Рус 23 -2019/06/03 14:04

ВОПРОСЫ И ОТВЕТЫ ОТНОСИТЕЛЬНО АРТЕРИАЛЬНОГО ДАВЛЕНИЯ
Q Почему артериальное давление, измеренное дома, отличается отартериального давления, измеренного в больнице?

A Артериальное давление зависит от физического состояния,температуры окружающей среды и психического состояния. Когда
врач или медсестра измеряет артериальное давление, показания
давления, скорее всего, будут выше обычного на 10-20 mmHg
вследствие взволнованности и/или нервозности пациента. Знание
факторов, влияющих на артериальное давление, поможет вамлучше следить за состоянием вашего здоровья.

Q Почему измеренное артериальное давление отличается прикаждом измерении?

Pyсский
CH-650̲1901.indb 24

A Наше артериальное давление автоматически регулируется

нервной системой. Артериальное давление отличается прикаждом ударе сердца. Мы думаем, что наше давление является
постоянным, однако оно изменяется, даже если проводить
измерения повторно. Артериальное давление чувствительно квремени дня, времени года и температуре воздуха. Кроме того,на артериальное давление оказывают влияние психические
стрессы, эмоциональные подъемы и спады. Оно увеличивается,когда вы напрягаетесь, и снижается, когда расслабляетесь.

Q Какие преимущества дает измерение артериального давления дома?
A Измерение артериального давления домадает более достоверные результаты, таккак вы можете выполнять измерение, когда
находитесь в спокойном и расслабленном
состоянии. Врачи придают большое значение
показаниям замеров артериального давленияв домашних условиях. Следует измерять
артериальное давление в одно и то же время
каждый день и вести записи изменений впоказаниях замеров артериального давления.

Всегда обращайтесь к врачу за консультацией по поводу
анализа результатов измерения артериального давления иназначением требуемого лечения.

- Рус 24 -2019/06/03 14:04

ПРЕЖДЕ ЧЕМ ОБРАЗАТЬСЯ ЗА РЕМОНТОМ ИЛИ ПРОВЕРКОЙ
Перед тем как отправить прибор для измерения артериального
давления на проверку или для ремонта, проверьте следующее.

Приборне может
измерить
артериальное
давление.

Проверьте

Решение

Проверьте, неразрядились либатарейки.

Замените все четыре
батарейки на новые.

Проверьте
правильность
полярности
установленных
батареек
(правильность
ориентации иУстановите батарейки,соблюдая правильную
полярность.

).

Проверьте,отображается лизначо .

Наденьте манжету
правильно.

Pyсский

ПроблемаПри нажатии
кнопки
“START/STOP”
экран дисплея
остается
пустым.

Проверьте,правильно линадета манжета.
Находились ли выво время измеренияв состоянии покоя?

Повторите измерение.
Не двигайтесь во время
измерения.

Прибор может не суметь выполнить
измерение артериального давления у лицсо слишком слабым пульсом или сердечной
аритмией (нерегулярным пульсом).
Показания замеров артериального
давления ненормально высокие или низкиепо сравнению с полученными в больнице.

Вытащите батарейки иустановите новые.

Другие проблемы

CH-650̲1901.indb 25

См. стр. 24- Рус 25 -2019/06/03 14:04

ТЕХНИЧЕСКИЕ ХАРАКТЕРИСТИКИ
Модель
Система измерения
Дисплей
Диапазон обхвата запястья
Диапазон
Давление
измерения
ПульсДавление
Погрешность
ПульсНагнетание
Номинальное напряжение
Выпуск

Pyсский
CH-650̲1901.indb 26

Источник питания
Срок службы батареек
Температура
Относительная
влажность
Температура
Условия
Относительная
хранения
влажность
Функция памяти
Защита от поражения
электрическим током
Степень охраны
Режим работы
Защита от проникновения водыРазмеры
ВесСрок службы
Условия
эксплуатации

CH-650Осциллометрический метод
Цифровой дисплей
13,5 - 21,5 см
0 - 280 mmHg (мм. рт. ст.)40 - 180 ударов/мин
±3 mmHg (мм. рт. ст.)±5% от показаний
Автоматическое нагнетание
встроенным насосом
3 В пост. тока(: постоянный ток)Электромагнитный клапан быстрого
выпуска
2 батарейки размера AAA (LR03)(180 mmHg, 1 разЛриблиз 400 разв день, 22°C)
От 10°C до 40°C
От 15% до 85%
От -20°C до 60°C
От 10% до 95%
До 60 измерений
Внутренний источник питания

Используются детали типа BF
Непрерывная работа
IPX065(Ш) x 74(Д) x 29(В) ммПриблиз. 103 г (баз батарейки)5 летКомплект включает 2 батареи размера
ПринадлежностиAAA (LR03) для прибора, инструкциюпо эксплуатации, футляр для хранения
* Рабочей частью данного устройства является манжета.
* Диапазон барометрического давления от 700 гПа до 1060 гПа*Т очность данного устройства соответствует требованиям пункта 7.9
стандарта EN1060-3.
EN1060-3: Неинвазивные сфигмоманометры. Часть 3. (Европейский
стандарт)- Рус 26 -2019/06/03 14:04

CH-650̲1901.indb 27

MEMO

2019/06/03 14:04

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 28

2019/06/03 14:04

MANUAL DE INSTRUCCIONESPARA ELMONITOR DIGITAL DE PRESIÓN SANGUÍNEA

CH-650̲1901.indb 1

Español

CH-650Español

1901

2019/06/03 14:04

Contenidos
3 - OBSERVACIONES GENERALES
5 - PRECAUCIONES PARA EL USO Y MANTENIMIENTO
10 - NOMBRES Y LOCALIZACIÓN DE LOS COMPONENTES
12 - CÓMO INSTALAR LAS PILAS
13 - CÓMO REEMPLAZAR LAS PILAS
14 - CÓMO COLOCAR LA MUÑEQUERA
16 - CONSEJOS PARA CONSEGUIR MEDICIONES PRECISAS
18 - CÓMO MEDIR LA PRESIÓN ARTERIAL
20 - RECUPERACIÓN DE LOS DATOS ALMACENADOS EN LA
MEMORIA22 - INDICACIONES DE ERROR
23 - ACERCA DE LA “PRESIÓN SANGUÍNEA”

Español
CH-650̲1901.indb 2

24 - PREGUNTAS Y RESPUESTAS SOBRE LA PRESIÓNSANGUÍNEA
25 - ANTES DE SOLICITAR REPARACIONES O PRUEBAS
26 - ESPECIFICACIONES

Asegúrese de que tiene todos estos componentes

Monitor de
Estuche depresión sanguínea almacenamiento(para el monitor)
2 pilas AAA

Manual de
instrucciones- Esp 2 -2019/06/03 14:04

OBSERVACIONES GENERALES
1. Para desinflar el brazalete rápidamente, presione el interruptor “START/STOP”.
2. El brazalete incluido es apto para una circunferencia de muñeca de 13.5
a 21.5 cm.
3. Al colocar el manguito, el cuerpo del monitor de presión sanguínea
debería ser colocado en la parte interior de su muñeca.
4. El monitor no está destinado para uso como un equipo de diagnóstico,está destinado para uso en el hogar.
5. ¡La automedición no es una terapia! En ningún caso deberá usted
cambiar la dosis de medicamentos recetada por su médico.
6. Como preparación para la medición de la presión sanguínea, es
aconsejable que orine y después permanezca relativamente quieto
durante los 10 ó 15 minutos previos a la medición.
7. Hacer ejercicio, comer, beber, fumar, etc., antes de la medición puede
afectar los resultados.
8. La presión sanguínea varía constantemente a lo largo del día. La
medición de la presión sanguínea deberá efectuarse regularmente a lamisma hora cada día.
9. NO se impresione demasiado por los resultados de una sola medición.
Mantenga un registro de las variaciones de la presión sanguínea. Una
imagen completa solamente puede obtenerse luego de una grancantidad de lecturas.
10. El estrés emocional suele causar una elevación de la presión sanguínea.

Español

Observaciones generales sobre la presión sanguínea yla medición de la presión sanguínea

Para una correcta medición de la presión sanguínea

1. Respire profundamente cinco o seis veces y seguidamente relájese
antes de medir la presión sanguínea. Si se encuentra tenso durante la
medición de la presión sanguínea, no conseguirá una lectura válida.
2. La presión sanguínea se elevará si se encuentra en una situación depreocupación o de enfado, de falta de sueño o estreñimiento, o despuésdel ejercicio o una comida.
3. Envuelva el manguito alrededor de su muñeca cómodamente
ver pág. 14).
(
4. NO mida la presión sanguínea después de bañarse o beber.
5. Cuando sienta necesidad de orinar, hágalo antes de medir su presión
sanguínea.
6. Mida su presión sanguínea cuando la temperatura de la habitación estéalrededor de los 20ºC. NO mida su presión sanguínea cuando hagademasiado frío (por debajo de 10ºC) o demasiado calor (por encima de40ºC) en la habitación.
7. NO intente medir su presión sanguínea inmediatamente después debeber café o té o de fumar.

CH-650̲1901.indb 3- Esp 3 -2019/06/03 14:04

OBSERVACIONES GENERALES
8. Mida la presión sanguínea cuando esté tranquilo y relajado. Mantenga el centrodel brazalete a la altura de su corazón y NO mueva el brazo o hable.
9. NO mida la presión sanguínea después que el brazalete haya estado ensu muñeca durante unos minutos. Durante ese tiempo, su muñeca habrá
desarrollado una presión sanguínea mayor y no obtendrá una lectura válida.
10. Si se visualiza el Indicador de Movimiento del Cuerpo o el Indicador de
Latido Irregular (IHB), relájese y mida nuevamente su presión arterial.
Si el Indicador de Latido Irregular (IHB) aparece frecuentemente, se lerecomienda que consulte con un médico acerca de su estado de salud.
( Vea la página 19)
11. La siguiente ilustración muestra la clasificación de la presión sanguíneade acuerdo con las normas de la OMS.
Sistólica (mmHg)
180

Hipertensión grado 3 (grave)
Hipertensión grado 2 (moderada)

Español
CH-650̲1901.indb 4

Observaciones:
El gráfico no es exacto,
pero se puede utilizar comoorientación para comprenderlas mediciones de presión
sanguínea no invasivas.
El aparato está destinado
solamente para uso con
adultos.

160
Hipertensión Grado 1 (Leve)
140
130
120

Alta Normal
NormalÓptima80 85 90

100

110

Diastólica (mmHg)

Definiciones y clasificaciones de los niveles de presión sanguínea
12. Uso pretendido• Este dispositivo es un monitor de presión sanguínea no invasivo
mediante método oscilométrico. Este dispositivo puede medir la presión
sanguínea sistólica (SYS), la presión sanguínea diastólica (DIA) y el pulso
de forma automática.
• Este dispositivo está diseñado para utilizarlo para comprobar elestado de salud personal en casa, sin las indicaciones de un médico,
y no está pensado como dispositivo de diagnóstico.
• Este dispositivo no está diseñado para utilizarlo con recién nacidos oniños pequeños, ni tampoco para mediciones cíclicas automáticas.

- Esp 4 -2019/06/03 14:04

OBSERVACIONES GENERALES

Mida su presión sanguínea a la misma hora cada día.

* La presión sanguínea cambia a cada momento. Esto significa que losdatos recogidos durante un largo periodo de tiempo tienen una mayor
significación que los datos de una única medición. Por esta razón, debe
medir la presión sanguínea diariamente. Idealmente, debería medir su
presión sanguínea a la misma hora cada día siempre que sea posible.

PRECAUCIONES PARA EL USO Y MANTENIMIENTO
1. Si sufre de enfermedad cardiaca, presión sanguínea alta u otra
enfermedad circulatoria, antes de utilizar el monitor, consulte con su
médico.
2. Si siente una presión anormal de la muñequera o cualquier otra
irregularidad durante el uso del dispositivo, reduzca la presión
apagándolo inmediatamente o quitando las pilas y consulte en elestablecimiento donde adquirió el aparato.
3. Si piensa que el valor medido es anormal o si la medición le hacesentirse indispuesto, interrumpa el uso y consulte con su médico.
4. La medición de la presión sanguínea puede no ser posible en
alguien con un pulso débil o arritmia.
5. Si la presión sanguínea se mide repetidamente, en algunas
personas puede causar problemas como congestión o hinchazón.
6. Las repeticiones frecuentes de las mediciones de la presión
sanguínea no darán resultados precisos. Deje un intervalo de 1
minuto entre las mediciones.
7. Consulte a su médico antes de usar el monitor si sufre usted deuna alteración severa de la circulación sanguínea en los brazos. El
no hacerlo puede conllevar riesgo para la salud.
8. La medición puede no ser posible en alguien con un flujo sanguíneo
insuficiente en el área donde se harán las mediciones o que sufre
de frecuentes latidos irregulares. Consulte con su médico paraconsejo sobre si usar o no el monitor.
9. NO envuelva el brazalete alrededor de una muñeca herida.
10. NO envuelva el brazalete alrededor de una muñeca en el que hayasido colocado un gotero (infusión intravenosa) o que esté siendoCH-650̲1901.indb 5

Español

Precauciones para el uso- Esp 5 -2019/06/03 14:04

PRECAUCIONES PARA EL USO Y MANTENIMIENTO

Español
CH-650̲1901.indb 6

usada para transfusión sanguínea como parte de un tratamiento
médico. El hacerlo puede causar heridas o un accidente grave.
11. NO opere en la vecindad de gases inflamables como los utilizados
para anestesia. Hacer esto podría encender los gases y causar una
explosión.
12. NO utilice el monitor en ambientes de oxígeno enriquecido, comoen una cámara hiperbárica de hospital o una tienda de oxígeno.
Hacer esto podría encender el oxígeno y causar un incendio.
13. NO use un teléfono móvil cerca del monitor porque esto podría
causar mal funcionamiento.
14. Si usa un marcapasos cardíaco, consulte con su medico antes deusar el monitor.
15. Asegúrese de utilizar esta unidad solamente para la medición de lapresión sanguínea. NO la utilice para ningún otro propósito.
16. Asegúrese de usar sólo piezas y accesorios originales autorizados.
Las piezas y los accesorios no aprobados para uso con el equipo
pueden dañar la unidad.
17. NO utilice esta unidad en niños.
18. Es posible que no se pueda realizar la medición de la presión
sanguínea en personas con arritmias frecuentes tales como latidos
auriculares o ventriculares prematuros o fibrilación auricular.

Precauciones para el mantenimiento
1. NO almacene el monitor de presión sanguínea donde vaya a estar
expuesto a la luz directa del sol, temperaturas elevadas (por encima
de 60ºC), temperaturas bajas (por debajo de -20ºC), humedad
relativa elevada (por encima de 95%) o en un sitio polvoriento.
Asegúrese de guardar el monitor de presión sanguínea fuera delalcance de niños, mascotas y pesticidas.
2. NO deje caer o exponga el monitor de presión sanguínea a
impactos o vibración.
3. Quite las pilas cuando el monitor no se vaya a utilizar durante unperiodo largo.
4. NO intente desarmar el monitor.
5. NO doble excesivamente el brazalete.
6. Cuando el monitor esté muy sucio, límpielo con un paño
humedecido con alcohol esterilizante o un detergente neutro.
Límpielo después con un paño seco.

- Esp 6 -2019/06/03 14:04

PRECAUCIONES PARA EL USO Y MANTENIMIENTO
7. NUNCA limpie el monitor de presión sanguínea con disolvente obenceno porque esto podría dañar el monitor.
8. Para limpiar el brazalete, límpielo con un paño húmedo. Evite frotar
con fuerza ya que esto puede causar fugas de aire.

Advertencia

No se permite la modificación de este
dispositivo.

CH-650̲1901.indb 7: La marca CE está diseñada para proporcionar informacióna los inspectores de mercado de los países miembros dela Unión Europea.
: Pieza aplicada tipo BF: Mantener seco
: Aviso: Consultar el manual de instrucciones antes del uso: Directiva WEEE de cumplimiento del dispositivo: Fabricante

Advertencia

El dispositivo ha sido comprobado yhomologado de acuerdo con EN60601-1-2
para EMC. Esto no garantiza de ninguna
forma que el dispositivo no estará afectado
por interferencia electromagnética. Evite
utilizar el dispositivo en un entorno altamente
electromagnético.

Español

Explicación de los símbolos- Esp 7 -2019/06/03 14:04

PRECAUCIONES PARA EL USO Y MANTENIMIENTONaturaleza y frecuencia del mantenimiento:

Este producto está diseñado para su uso durante un largo período de tiempo; sinembargo, es generalmente recomendado que sea inspeccionado cada dosaños
para asegurar su funcionamiento y rendimiento correctos.

Protección del medioambiente:

Al deshacerse de la pila y del producto al final de su vida útil contribuya a proteger
el medioambiente respetando las correspondientes regulaciones nacionales ylocales sobre reciclado.

MARCO DE RAEESi desea deshacerse de este producto, no lo mezcle con residuos
domésticos de carácter general. Existe un sistema de recogida selectiva
de aparatos electrónicos usados, según establece la legislación prevista
por la Directiva 2012/19/EU sobre residuos de aparatos eléctricos yelectrónicos (RAEE), vigente únicamente en la Unión Europea.

Información sobre la eliminación en otros países fuera de la
Unión Europea.

Si desea desechar pilas usadas, póngase en contacto con las autoridades localeso pregunte cuál es el método de eliminación adecuado.

Español
CH-650̲1901.indb 8

Nota sobre el símbolo de pila

El símbolo se puede usar en combinación con un símbolo químico. En este caso,
cumple los requisitos establecidos por la Directiva de la sustancia química en
cuestión.

Estándar Europeo de Referencia:

El equipo de medición de presión sanguínea está en conformidad con lasregulaciones EN60601-1, EN1060-1, EN1060-3, EN1060-4.

Calibración:

Se recomienda inspeccionar el monitor de equipo de medición de presión
sanguínea cada dos años para garantizar su correcto funcionamiento yprecisión. Contacte con un distribuidor.

- Esp 8 -2019/06/03 14:04

PRECAUCIONES PARA EL USO Y MANTENIMIENTO

Nombre :Dirección :Fábrica

Nombre :Dirección :CITIZEN SYSTEMS JAPAN CO.,LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan
CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen,
Guangdong, China

Representante europeo:

Nombre :Dirección :EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague,
The Netherlands•P ara obtener datos técnicos, póngase en contacto

con nuestro representante europeo.

CH-650̲1901.indb 9

EspañolHecho en China- Esp 9 -2019/06/03 14:04

NOMBRES Y LOCALIZACIÓN DE LOS COMPONENTES
Nombres de los componentes<Unidad principal>

PantallaInterruptor MEMORY

EspañolInterruptor
START/STOP
EsperaCH-650̲1901.indb 10

Tapa delcompartimientode las pilas

Brazalete de muñeca- Esp 10 -2019/06/03 14:04

NOMBRES Y LOCALIZACIÓN DE LOS COMPONENTES<Pantalla>

Presión sanguínea
sistólica

Indicador de latido
irregular (IHB)
Pulso

Indicador demovimiento del cuerpo

CH-650̲1901.indb 11

EspañolPresión sanguínea
diastólica

Indicador del nivel de las pilas
Aparece cuando las pilas están por
Vea la página 13)
agotarse. (- Esp 11 -2019/06/03 14:04

CÓMO INSTALAR LAS PILAS(Las pilas incluidas con el aparato deben introducirse en su
compartimiento antes de poner el esfigmomanómetro en uso.)

1 Abrir la tapa del compartimiento de las pilas.

Tapa del compartimientode las pilas

Español
CH-650̲1901.indb 12

Presionando la tapa delcompartimiento de las pilas,
deslice la tapa hacia fuera enla dirección de la flecha.

2 Colocar las pilas en el compartimiento de las pilas.
Colocar las pilas siguiendo las indicaciones de los símbolos* La pila entra primero con el lado negativo .

y.

Instale la pila del ladode la pantalla primero.

- Esp 12 -2019/06/03 14:04

CÓMO INSTALAR LAS PILAS

3 Cerrar la tapa del compartimiento de las pilas.
Deslice la tapa en la direcciónde la flecha. Empuje la tapa
hasta que se encaje.

• Cuando se visualiza el icono
o cuando no aparece nada en lapantalla, reemplazar las pilas por dos pilas nuevas. Cambiar las dos
pilas al mismo tiempo.
• Utilice siempre pilas alcalinas.
• No utilice pilas recargables.

EspañolCÓMO REEMPLAZAR LAS PILAS• No utilice pilas alcalinas (LR03) y de manganeso (R03) juntas.
• Las pilas incluidas son tan sólo para demostración. La vida de la pila
podría ser menor de lo especificado.
• Cuando deseche pilas usadas, cumpla las normativas estatales olas normas de la institución pública medioambiental aplicables en su
país/área.

CH-650̲1901.indb 13- Esp 13 -2019/06/03 14:04

CÓMO COLOCAR LA MUÑEQUERA
Envuelva la muñequera alrededor de su muñeca correctamente. Una
envoltura incorrecta podría repercutir en errores de medida.

1 Abra la muñequera.
2 Envuelva la muñequera aIrededor

de su muñeca izquierda de manera
que el monitor está posicionado enla parte interior de su muñeca.
• Coloque la muñequera sobre pieldesnuda.
• La muñequera está diseñada para
ajustarse a muñecas de 13.5 a
21.5 cm de circunferencia.

Español
CH-650̲1901.indb 14

3 Holgadamente ajuste lamuñequera tirando delextremo de la misma.

- Esp 14 -2019/06/03 14:04

CÓMO COLOCAR LA MUÑEQUERA

4 Posicione el borde superior de lamuñequera aproximadamente de 1a 2 cm de distancia del pliegue desu muñeca.

5 Ajuste la muñequera con elsistema de cierre gancho y bucle.

Español• Si la muñequera está
colocada demasiado
holgada o demasiado
tirante, podría resultar
en errores o lecturas
incorrectas.

Puede usar su muñeca derecha para medir su presión
sanguínea.
Sin embargo, puede que la presión sanguínea difiera aproximadamente
10 mmHg dependiendo si se mide en la muñeca izquierda o derecha y,por lo tanto, idealmente se debe medir la presión sanguínea siempre enla misma muñeca.
Para obtener mediciones precisas• Un brazalete muy flojo o muy apretado puede impedir la medición
correcta.
• Siéntese derecho mientras toma la medición.

CH-650̲1901.indb 15- Esp 15 -2019/06/03 14:04

CONSEJOS PARA CONSEGUIR MEDICIONES PRECISAS
La presión sanguínea varía con su postura, la hora del día y una gamade factores. Idealmente, debe medir su presión sanguínea a la misma
hora cada día en la misma postura.

➀ Asegúrese de que su postura esté correcta.

Siéntese derecho, respire profundamente y relaje.

➁ Coloque la palma de la mano
hacia arriba y deje los dedos
relajados.

➂ Es importante colocar

el brazalete al nivel delcorazón.

Español
CH-650̲1901.indb 16

Coloque una toalla doblada oalgo similar por debajo de su
brazo.

- Esp 16 -2019/06/03 14:04

CONSEJOS PARA CONSEGUIR MEDICIONES PRECISAS

Para obtener mediciones correctas

CH-650̲1901.indb 17• No su apriete el puño con fuerza.

• No cambie la posición del monitor.

Español• Envuelva el brazalete sin dejarlo muy
flojo.

• No mueva el cuerpo ni hable durante la
medición.

- Esp 17 -2019/06/03 14:04

CÓMO MEDIR LA PRESIÓN ARTERIAL
1 Coloque la muñequera alrededor de su muñeca izquierda.
(

Ver página 14 “CÓMO COLOCAR LA MUÑEQUERA”.)

2 Asegúrese de que su postura es correcta.
(

 er página 16 “CONSEJOS PARA CONSEGUIR MEDICIONESVPRECISAS”.)

3 Pulse el interruptor “START/STOP” para

iniciar la medición.
• La visualización de todos los dígitos es
una pantalla inicial, que indica que elmonitor está funcionando correctamente.
La marca
también aparece, pero esto
no indica que se requiere reemplazar la
pila.

Español
CH-650̲1901.indb 18

Se visualiza

cuando se detecta el pulso.

- Esp 18 -2019/06/03 14:04

CÓMO MEDIR LA PRESIÓN ARTERIAL

4 Se visualizan los resultados de la medición.
Presión sanguínea
sistólica

Presión sanguínea
diastólica
Indicador de latido
irregular (IHB)

CH-650̲1901.indb 19

Pulso

El indicador de movimiento del cuerpo aparece enla visualización de los resultados de la medición simueve su mano o muñeca durante la medición,lo que causa un cambio sustancial de la presión adetectarse.
• Si se visualiza el indicador de movimiento delcuerpo, mida de nuevo la presión sanguínea.

Español

Indicador de movimiento del cuerpo

Si se detecta un latido irregular durante la medición,el indicador de latido irregular (IHB) aparece una vez
que se concluya la medición.
Un latido irregular es un latido que varía ±25%
desde el medio de intervalos de 5 latidos durante la
medición de la presión sanguínea.
• Puede que la medición no sea correcta si su pulso
fluctúa grandemente durante la medición. Si sevisualiza el indicador de latido irregular (IHB), mida
de nuevo la presión sanguínea.

Precaución

Si el indicador de latido irregular (IHB) aparece
con frecuencia, se recomienda que consulte
un médico sobre su condición de salud.

- Esp 19 -2019/06/03 14:04

CÓMO MEDIR LA PRESIÓN ARTERIAL

5 Terminando la medición.
• Pulsar “START/STOP” para apagará el monitor.

FUNCIÓN DE APAGADO AUTOMÁTICO• Después de tomar la medición de la presión arterial, pulsando elinterruptor “START/STOP” se apagará el monitor. Sin embargo, si no
pulsa el interruptor, la función de apagado automático del monitor
apagará automáticamente el monitor después de 3 minutos.

RECUPERACIÓN DE LOS DATOS ALMACENADOS EN LA MEMORIA
60 resultados de medición pueden almacenarse en la memoria. El valor
promedio se calcula automáticamente para ayudarle a administrar su
salud diaria.

Español
CH-650̲1901.indb 20CÓMO RECUPERAR LOS DATOS
1 Presione el interruptor “MEMORY”.

• En la parte izquierda de la pantalla aparece “3A” indicando elvalor medio. Luego se muestra el valor medio de las tres últimas
mediciones (“3A” y el valor medio no aparecen si hay menos de dos
mediciones almacenadas en la memoria).

- Esp 20 -2019/06/03 14:04

RECUPERACIÓN DE LOS DATOS ALMACENADOS EN LA MEMORIA

2 Cada vez que se pulsa el interruptor, se visualizan los datos demedición en el orden de los más recientes a los más antiguos.
Se visualiza el número
de los datos antes delos resultados de la
medición.

• Los valores guardados en la memoria están en el orden de las
mediciones. Por ejemplo, cuando se han guardado 60 mediciones,
la número 1 representa la última medición realizada.
Número de datos: 1, 2, 3, ... ... 59, 60ÇÇLa última medición
La primera medición

Español

Se visualizan los resultados de la medición después del número de los datos.

CÓMO BORRAR LOS DATOS
1 Pulse el interruptor “MEMORY”.
2 Mientras pulsa el interruptor “MEMORY”, pulse el interruptor“START/STOP” por más de 3 segundos.
Se borrarán todos los datos almacenados.

CH-650̲1901.indb 21

Pulse por 3
segundos

Borrados

Se visualiza
un mensaje
informando que
todos los datos
han sido borradosde la memoria.

- Esp 21 -2019/06/03 14:04

INDICACIONES DE ERRORSEÑAL DE VISUALIZACIÓN

Español
CH-650̲1901.indb 22

CONDICIÓN/CAUSAACCIÓN CORRECTORA

Aparece cuando
la muñequera está
colocada demasiado
holgadamente.

Recoloque la muñequera
cómodamente. Tome otra
lectura, asegurándose deque permanece quieto.
(
Ver página 14“CÓMO COLOCAR LAMUÑEQUERA”.)

Aparece cuando su
muñeca se ha movido
durante la toma, oaparece cuando elmanguito no está
colocado correctamente.

Recoloque la muñequeracorrectamente. Tome otra
lectura asegurándosede permanecer
perfectamente quieto.
(ver página 14“CÓMO COLOCAR LAMUÑEQUERA”.)

Aparece cuando las pilas
se han agotado.

Reemplace las dos
pilas alcalinas por
unas nuevas. (Ver página 13 “CÓMOREEMPLAZAR LASPILAS”.)

Aparece cuando el ritmo
de pulsaciones está fuera
del rango de medida (pordebajo de 39 latidos/min.
o por encima de 181
latidos/min.)Recoloque la muñequeracorrectamente. Inspire
varias veces profundamentepara relajarse, y mida denuevo. (Ver página
16 “CONSEJOS PARACONSEGUIR MEDICIONESPRECISAS”.)

La unidad no funciona
correctamente.

Contacte con la tienda
donde fue comprada ocon su centro local deservicio. (Ver página
25 “ANTES DE SOLICITARREPARACIONES OPRUEBAS”.)- Esp 22 -2019/06/03 14:04

ACERCA DE LA “PRESIÓN SANGUÍNEA”¿Qué es la presión sanguínea?

El corazón es una bomba que impulsa la sangre por todo su cuerpo. La sangre esbombeada a presión constante desde el corazón a las arterias. A esta presión se lellama la presión sanguínea arterial, y representa, en términos generales, su presión
sanguínea. La presión sanguínea está indicada por diferentes tipos de presiones,
entre ellas la presión sistólica que ocurre cuando el corazón bombea sangre, y lapresión diastólica que ocurre cuando la sangre regresa al corazón.
(Sistólica)(Diastólica)
Arteria

La pared arterial
se expande.

Ventrículo
izquierdo

Contracción

Se restaura el estado
normal en la pared arterial.
Dilatación

La presión arterial varía a toda hora.

Su presión sanguínea cambia con su edad, sexo y una gama de otros factores. Es probableque resulte afectada por su biorritmo durante el día y por su postura, actividades físicas,
actividades intelectuales, el nivel de estrés e incluso por la temperatura atmosférica. La
presión sanguínea de una persona saludable generalmente varía a lo largo de un día.

Español

Ventrículo
izquierdo

ArteriaAtrio
izquierdo

Atrio
izquierdo

Valor de la presión sanguínea (mmHg)[Ejemplo de las variaciones de la presión sanguínea durante un día]
Presión sanguínea
sistólica

160
140
120
100
80
60
40

Sueño

Presión sanguínea
diastólica

12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11

Hora del día(Valores de presión sanguínea medidos en intervalos de cinco minutos durante un día)l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329–44.

CH-650̲1901.indb 23- Esp 23 -2019/06/03 14:04

PREGUNTAS Y RESPUESTAS SOBRE LA PRESIÓN SANGUÍNEA
P ¿Por qué la medición de la presión sanguínea en casa es diferente de lamedida por el médico o en un hospital?

R La presión sanguínea puede variar con el ejercicio, la temperatura ambientalo su estado de ánimo. Cuando un médico o enfermera toma su presión
sanguínea, el valor de la presión probablemente será 10 o 20 mmHg más altode lo normal, debido a la preocupación y al estrés. Sabiendo qué es lo que
afecta a su presión sanguínea le ayudará a la mejor monitorización su salud.

P ¿Por qué los resultados de la medición de la presión arterial varían con
cada medición?

R Nuestra presión arterial la ajusta un nervio automáticamente. La presión

Español
CH-650̲1901.indb 24

arterial difiere con cada latido del corazón. Pensamos que nuestra
presión arterial es constante pero se puede ve como varía al realizar
varias mediciones seguidas. La presión arterial es diferente a diferentes
horas del día, tales como la mañana y la tarde, en diferentes estaciones
y en diferentes temperaturas atmosféricas. Además, la presión arterial seve fácilmente afectada por el estrés o los cambios emocionales. Tiende aaumentar cuando se siente tensión y a disminuir cuando se está relajado.

P ¿Cuál es el beneficio de medir la presión sanguínea en casa?
R La medición de la presión sanguínea en casa es fiable puesto que puede
ser realizada en una situación estable y relajada. Los doctores dan
mucha importancia a las mediciones de presión sanguínea tomadas en
casa. Se recomienda medir su presión sanguínea a la misma hora cadadía y guardar las variaciones de su presión sanguínea.

Siempre consulte con su médico para la interpretación delas lecturas de su presión sanguínea, y para determinar el
tratamiento correcto.

- Esp 24 -2019/06/03 14:04

ANTES DE SOLICITAR REPARACIONES O PRUEBAS
Asegúrese de comprobar los siguientes puntos antes de enviar su
monitor de presión sanguínea a ser probado o reparado.
Problema

La pantalla nomuestra nadaaunque pulse
el interruptor“START/STOP”.

No se puede
medir.

Comprobaciones

Respuesta

Compruebe silas pilas están
agotadas.

Cambie todas las pilas
por unas nuevas.

Compruebe silas pilas estáncorrectamente
instaladas ( y enla posición correcta).

Instale las pilas en la
dirección correcta.

Compruebe si haaparecido el icono.

Coloque el brazaletecorrectamente.

¿Permaneció
tranquilo durante la
medición?

EspañolCompruebe
si el brazaleteestá colocado
correctamente.
Vuelva a medir su presión
sanguínea, asegurándose
de no moverse.

Tenga en cuenta que la medición de la presión
sanguínea pudiera no ser posible para alguien
con un pulso extremadamente débil o arritmia
cardiaca (pulsos irregulares).
Sus lecturas de presión sanguínea sonanormalmente altas o bajas respecto alas medidas en el hospital.
Otros fenómenos

CH-650̲1901.indb 25

Vea la página 24.

Retirar las pilas delaparato y colocarle otras
nuevas.

- Esp 25 -2019/06/03 14:04

ESPECIFICACIONES
Número de modelo
Sistema de medición
Pantalla
Gama de circunferencia demuñecaGama dePresión
medición
Pulso
Pressure
Accuracy
Pulso
InfladoVoltaje nominal
EscapeFuente de alimentación
Duración de las pilas

Español
CH-650̲1901.indb 26

Condiciones
Temperaturade operación Humedad
Temperatura
Storageconditions
Humedad
Función de memoria
Protección contra descarga
eléctrica
Grado de protección
Modo de funcionamiento
Protección contra el ingresode aguaDimensiones
Peso
Vida útil
Accesorios

CH-650
Método oscilométrico
Tipo de pantalla digital
13.5 a 21.5 cm
0 a 280 mmHg
40 a 180 pulsaciones/min±3 mmHg±5% de la lectura
Inflado automático mediante bomba interna
3V DC(: Corriente Directa)
Válvula electromagnética de escape rápido
2 pilas AAA (LR03)(180 mmHg,
Aprox. 400 veces
una vez al día, 22°C)
10°C a 40°C
15% a 85% RH-20°C a 60°C
10% a 95%RH
Hasta 60 medidas
Unidad de alimentación interna
Pieza aplicada tipo BF
Funcionamiento continuo
IPX065(An) x 74(Al) x 29(Pr) mm
Aproximadamente 103g excluyendo pilas
5 añosJuego compuesto por 2 pilas AAA (LR03)
para el monitor, manual de instrucciones,
estuche de almacenamiento* La pieza aplicada para este dispositivo es el brazalete.
* Intervalo de presión barométrica: de 700 hPa a 1060 hPa* La precisión general del sistema de este dispositivo se cumple con elrequisito de la sección 7.9 de la norma EN1060-3.
EN1060-3 : Tensiómetros no invasivos, sección 3 (norma europea)- Esp 26 -2019/06/03 14:04

CH-650̲1901.indb 27

MEMO

2019/06/03 14:04

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 28

2019/06/03 14:04

MANUAL DE INSTRUÇÕES DO UTILIZADORDEMEDIDOR ELECTRÓNICO DIGITAL DA TENSÃO
ARTERIAL

CH-650̲1901.indb 1

Português

CH-650Português

1901

2019/06/03 14:04

Índice3 - OBSERVAÇÕES GERAIS
5 - CUIDADOS A TER NA UTILIZAÇÃO E MANUTENÇÃO
10 - NOMES E LOCALIZAÇÕES DOS COMPONENTES
12 - COLOCAR AS PILHAS
13 - RETIRAR AS PILHAS
14 - COMO ENROLAR A BRAÇADEIRA DE PULSO
16 - SUGESTÕES SOBRE COMO OBTER MEDIÇÕES EXACTAS
18 - MEDIR A TENSÃO ARTERIAL
20 - ACEDER AOS DADOS GUARDADOS NA MEMÓRIA
22 - INDICAÇÕES DE ERRO
23 - SOBRE A TENSÃO ARTERIAL
24 - PERGUNTAS E RESPOSTAS SOBRE A TENSÃO ARTERIAL
25 - ANTES DE SOLICITAR REPARAÇÕES OU INSPECÇÃO
26 - ESPECIFICAÇÕES

Português
CH-650̲1901.indb 2

Assegure-se de que possui todos os seguintes componentes

Medidor datensão arterial

Estojo dearrumação

(para o medidor)
2 pilhas detamanho “AAA”

Manual deInstruções

- Por 2 -2019/06/03 14:04

OBSERVAÇÕES GERAIS
Observações gerais sobre a tensão arterial e sua medição

1. Para desinsuflar a braçadeira rapidamente, prima o botão “START/STOP”.
2. A braçadeira fornecida é adequada para um pulso com um
perímetro entre 13,5 e 21,5 cm.
3. Quando colocar a braçadeira, o corpo do monitor da pressão
arterial deve ser colocado na parte interior do seu pulso.
4. O medidor foi concebido para utilização doméstica e por isso não
visa ser um equipamento de diagnóstico.
5. A automedição não é uma terapia! Não deverá, em circunstância
alguma, alterar a dosagem da medicação prescrita pelo seu médico.
6. Antes de medir a tensão arterial deverá urinar e depois permanecer
relativamente quieto durante 10 a 15 minutos.
7. Fazer exercício, comer, beber, fumar, etc., antes de medir a tensão
arterial pode afectar os resultados.
8. A tensão arterial varia constantemente ao longo do dia. As medições
devem ser feitas regularmente todos os dias e sempre à mesma hora.
9. NÃO valorize demasiado os resultados de uma medição. Mantenhaum registo contínuo das variações da tensão arterial. Só um grandenúmero de leituras pode dar uma clara indicação da situação.
10. O stress emocional tende a fazer subir a tensão arterial.
1. Respire fundo cinco ou seis vezes e depois descontraia-se antes
de medir a tensão arterial. Se estiver tenso durante a medição datensão arterial o valor não será um valor válido.
2. A tensão arterial estará elevada se estiver preocupado ou irritado,
se não tiver dormido bem, se estiver com prisão de ventre, ou setiver acabado de comer ou de fazer exercício.
3. Enrole a braçadeira comodamente à volta do seu pulso. (Verpágina 14)
4. NÃO meça a tensão arterial depois de tomar banho ou de beber.
5. Se sentir necessidade de urinar, faça-o antes de medir a tensão arterial.
6. Meça a tensão arterial num local onde a temperatura ambiente
ronde os 20 °C. NÃO meça a tensão arterial quando estiver muito
frio (abaixo de 10 °C) ou muito calor (acima de 40 °C).
7. NÃO meça a tensão arterial logo após ter bebido café ou chá ou
após ter fumado.
8. Meça a tensão arterial quando estiver relaxado e quieto. Mantenhao centro da braçadeira ao nível do seu coração e NÃO mexa obraço nem fale.

CH-650̲1901.indb 3

Português

Para obter medições correctas da tensão arterial- Por 3 -2019/06/03 14:04

OBSERVAÇÕES GERAIS
9. NÃO meça a tensão arterial depois de a braçadeira estar colocada
há alguns minutos. Durante esse tempo, o seu pulso teráacumulado uma tensão arterial mais elevada e a leitura não seráválida.
10. Se o indicador de movimento do corpo ou de batimento cardíaco
irregular (IHB) for exibido, tente relaxar e meça a tensão arterial denovo. Se o indicador de batimento cardíaco irregular (IHB) surgir
com frequência, deve consultar um médico para verificar o seu
estado de saúde. (Consulte a página 19.)
11. A seguir podem ver-se as classificações da tensão arterial segundo
as normas da OMS.
Sistólica (mmHg)
180

Hipertensão de grau 3 (grave)
Hipertensão de grau 2 (moderada)
Observações:O gráfico não é exacto, maspode ser utilizado comoorientação para compreender
as medições da tensão arterial
não-invasiva. O dispositivo
destina-se apenas à medição
da tensão arterial em adultos.

160
Hipertensão de grau 1 (ligeira)
140

Português
CH-650̲1901.indb 4

130
120

Elevada-normal
NormalÓptima80 85 90

100

110

Diastólica (mmHg)

Definições e classificações dos níveis da tensão arterial
12. Uso pretendido• Este dispositivo é um medidor de tensão arterial não invasivo
que utiliza o método de oscilometria. Este dispositivo podemedir a pressão sanguínea sistólica (SYS), a pressão sanguínea
diastólica (DIA) e a pulsação automaticamente.
• Este dispositivo destina-se a ser utilizado para examinar acondição de saúde pessoal em casa, sob a direcção de um
médico, e não deve ser utilizado como um dispositivo dediagnóstico.
• Este dispositivo não foi concebido para utilização com recémnascidos e bebés.

- Por 4 -2019/06/03 14:04

OBSERVAÇÕES GERAIS

Meça a sua tensão arterial todos os dias sempre à
mesma hora.

* A tensão arterial está sempre a sofrer alterações. Isto significa que
os dados reunidos ao longo de um determinado período de tempo
têm mais significado do que uma só medição. É portanto necessário
medir a tensão arterial diariamente. O ideal seria medir a tensão
arterial todos os dias à mesma, se possível.

CUIDADOS A TER NA UTILIZAÇÃO E MANUTENÇÃO
1. Se sofrer de doença cardíaca, de tensão arterial elevada ou deoutras doenças do aparelho circulatório, consulte o seu médico
antes de utilizar o medidor.
2. Se sentir pressão anormal da braçadeira durante o uso ou qualquer
outra irregularidade, reduza a pressão desligando imediatamentea energia ou retire as pilhas e consulte o posto de venda onde
efetuou a compra.
3. Se achar que os valores da medição não são normais ou se as
medições o fizerem sentir-se mal, deixe de utilizar o medidor e
consulte o seu médico.
4. A medição da tensão arterial pode não ser possível em pessoas
com pulso fraco ou arritmia.
5. A medição da tensão arterial efectuada repetidamente pode causar
problemas em algumas pessoas, como exemplo inchaço ou
congestão sanguínea.
6. Medir a tensão arterial frequente e repetidamente não produzirá
resultados correctos. Deverá aguardar-se 1 minuto antes de voltar
a medir a tensão.
7. Consulte o seu médico antes de utilizar o medidor, se tiver qualquer
problema grave de circulação sanguínea nos braços. A não
observância desta instrução pode pôr a sua saúde em risco.
8. A medição pode não ser possível em pessoas que tenham um
fluxo sanguíneo insuficiente na área onde se vai medir a tensão
ou em pessoas que tenham pulsações irregulares frequentes.
Consulte o seu médico para saber se pode utilizar o medidor.

CH-650̲1901.indb 5

Português

Cuidados a ter na utilização

- Por 5 -2019/06/03 14:04

CUIDADOS A TER NA UTILIZAÇÃO E MANUTENÇÃO

Português
CH-650̲1901.indb 6

9. NÃO coloque a braçadeira num pulso ferido.
10. NÃO coloque a braçadeira num pulso que esteja a ser utilizado
para administrar perfusão intravenosa ou aplicar transfusão desangue. A não observância desta instrução pode resultar numalesão ou num acidente grave.
11. NÃO utilize o medidor nas imediações de gases inflamáveis, comopor exemplo gases usados em anestesia. A não observância
desta instrução pode causar a ignição dos gases e consequente
explosão.
12. NÃO utilize o medidor em ambientes ricos em oxigénio, com por
exemplo uma câmara hiperbárica de um hospital ou uma tenda deoxigénio. A não observância desta instrução pode causar a ignição
do oxigénio e consequente incêndio.
13. NÃO utilize telemóveis perto do aparelho, já que isso pode causar
mau funcionamento.
14. Se utilizar um pacemaker consulte o seu médico antes de utilizar omedidor.
15. Assegure-se de que utiliza este medidor apenas para medir atensão arterial. NÃO o utilize para qualquer outra finalidade.
16. Certifique-se de que utiliza apenas os nossos componentes
e acessórios autorizados. Os componentes e acessórios não
aprovados para utilização com o equipamento podem danificar a
unidade.
17. NÃO utilize este medidor em crianças de pouca idade.
18. A medição da tensão arterial pode não ser possível em pessoas
com arritmias comuns, tais como batimentos auriculares ou
ventriculares prematuros ou fibrilhação auricular.

Cuidados a ter na manutenção

1. NÃO guarde o medidor da tensão arterial em locais expostos a luz
solar directa, temperatura elevada (acima de 60 °C), temperatura
baixa (abaixo de -20 °C), humidade relativa elevada (acima de 95%)
ou em locais com muito pó.
Certifique-se de guardar o medidor de pressão arterial num lugar
fora do alcance de crianças, animais de estimação ou pragas.
2. NÃO deixe cair o medidor da tensão arterial, nem o submeta a
impactos ou vibração.
3. Retire as pilhas quando não for utilizar o medidor durante muito
tempo.

- Por 6 -2019/06/03 14:04

CUIDADOS A TER NA UTILIZAÇÃO E MANUTENÇÃO
4. NÃO tente desmontar o medidor.
5. NÃO a braçadeira excessivamente.
6. Se o medidor estiver muito sujo, limpe-o com um pano
humedecido em álcool esterilizado ou em detergente neutro. A
seguir, limpe-o com um pano seco.
7. NUNCA limpe o medidor da tensão arterial com diluentes ou
benzeno, já que esses produtos podem danificar o medidor.
8. Para limpar a braçadeira utilize um pano húmido. Evite esfregar
com força já que isso provocará fugas de ar.

Aviso

É proibido fazer qualquer modificação neste
equipamento.

CH-650̲1901.indb 7: A marca CE destina-se a fornecer informações aos
inspectores de mercado nos países membros da UE.
: Peça aplicada Tipo BF: Manter seco
: Aviso: Consulte o manual de instruções antes de utilizar: Directiva de conformidade de aparelhos WEEE
: Fabricante

Aviso

O dispositivo foi testado e homologado deacordo com a directiva EN60601-1-2 para
compatibilidade electromagnética. Isto não
garante que o dispositivo não seja afectado
por interferências electromagnéticas. Evite
utilizar o dispositivo em ambientes com
grandes cargas electromagnéticas.

Português

Explicação dos símbolos- Por 7 -2019/06/03 14:04

CUIDADOS A TER NA UTILIZAÇÃO E MANUTENÇÃO

Português
CH-650̲1901.indb 8

Natureza e frequência da manutenção:Este produto foi concebido para ser utilizado durante muito tempo; noentanto, recomenda-se normalmente que seja inspeccionado de dois em
dois anos para assegurar um funcionamento e desempenho adequados.
Proteja o ambiente natural:
Ajude a proteger o ambiente natural respeitando a regulamentação
nacional e/ou local sobre reciclagem quando eliminar a pilha e oproduto no final da respectiva vida útil.
SÍMBOLO REEESe quiser deitar fora este produto, não o misture com o lixo
comum. De acordo com a legislação que decorre da Directiva
REEE – Resíduos de Equipamentos Eléctricos e Electrónicos(2012/19/EU), existe um sistema de recolha separado para os
equipamentos electrónicos fora de uso, em vigor apenas naUnião Europeia.
Informações acerca da eliminação noutros países fora da União Europeia.
Se pretender eliminar baterias usadas, contacte as autoridades locais
ou o fornecedor e informe-se do método correcto de eliminação.
Nota para o símbolo da bateria.
O símbolo pode ser utilizado juntamente com um símbolo químico.
Neste caso, cumpre os requisitos definidos pela Directiva para oquímico em causa.

Norma Europeia de Referência:

O dispositivo de medição de tensão arterial cumpre as normas
EN60601-1, EN1060-1, EN1060-3, EN1060-4.
Calibragem:

O monitor do dispositivo de medição da tensão arterial deve ser inspeccionado
de dois em dois anos para assegurar a exactidão e o correcto funcionamento
do mesmo. Por favor contacte um distribuidor.

- Por 8 -2019/06/03 14:04

CUIDADOS A TER NA UTILIZAÇÃO E MANUTENÇÃO

Nome :
Endereço :Fábrica

Nome :
Endereço :CITIZEN SYSTEMS JAPAN CO.,LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan
CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen,
Guangdong, China

Representante Europeu:

Nome :
Endereço :EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague,
The Netherlands•P ara dados técnicos, contacte os nossos

representantes europeus.

CH-650̲1901.indb 9

Português

Fabricado na China- Por 9 -2019/06/03 14:04

NOMES E LOCALIZAÇÕES DOS COMPONENTES
Nomes dos componentes<Unidade principal>

Visor

Interruptor MEMORY(Memória)
Botão
START/STOP
Posição de espera

Português
CH-650̲1901.indb 10

Tampa docompartimento
das pilhas

Braçadeira de pulso- Por 10 -2019/06/03 14:05

NOMES E LOCALIZAÇÕES DOS COMPONENTES<Visor>Pressão sistólica

Pressão diastólica

Pulsação

Indicador de movimento
do corpo

CH-650̲1901.indb 11

Indicador do nível das pilhas
Aparece quando as pilhas está aConsulte a página 13)
esgotar-se. (Português

Indicador de pulsação
irregular (IHB)- Por 11 -2019/06/03 14:05

COLOCAR AS PILHAS(As pilhas fornecidas com o aparelho devem ser colocadas nocompartimento próprio antes de começar a utilizar o medidor datensão arterial.)

1 Abra a tampa do compartimento das pilhas.

Tampa docompartimento daspilhasPremindo a tampa docompartimento das pilhas,
deslize a tampa para fora
na direcção da seta.

2 Coloque as pilhas no compartimento.
Português
CH-650̲1901.indb 12

Coloque as pilhas prestando atenção aos símbolos
indicadores dos terminais positivo e negativo.
* A pilha entra primeiro no lado negativo .

eInstale a pilha do lado
do visor primeiro.

- Por 12 -2019/06/03 14:05

COLOCAR AS PILHAS

3 Feche a tampa do compartimento das pilhas.
Deslize a tampa na direcção daseta. Empurre a tampa até quese encaixe.

• Quando a indicação
aparecer no visor ou quando o visor estivercompletamente em branco, substitua as pilhas usadas por duas
pilhas novas. Sempre que substituir pilhas, deve substituir as duasde uma vez.
• Utilize somente pilhas alcalinas.
• Não utilize baterias recarregáveis.
• Não utilize pilhas alcalinas (LR03) e de manganês (R03) juntas.
• As pilhas incluidas têm somente a finalidade de demonstração. O
tempo de vida útil da pilha pode ser inferior ao especificado.
• Quando eliminar baterias usadas, obedeça às normas ou regras
governamentais ou de instituições públicas ambientais que seaplicam ao seu país/área.

CH-650̲1901.indb 13

Português

RETIRAR AS PILHAS- Por 13 -2019/06/03 14:05

COMO ENROLAR A BRAÇADEIRA DE PULSO
Enrole adequadamente a braçadeira de pulso à volta do seu pulso. O
enrolar inadequado da braçadeira pode originar erros de medição.

1 Abra a braçadeira de pulso.
2 Enrole a braçadeira à volta do seu

pulso esquerdo de forma a que omonitor fique posicionado na parteinterior do seu pulso.
• Enrole a braçadeira de pulso sobrea pele nua.
• A braçadeira de pulso estáconcebida para suportar pulsos
com uma circunferência que vádesde 13,5 a 21,5 cm.

3 Ajuste livremente a braçadeira de pulso ao braço puxando
pela extremidade da braçadeira.

Português
CH-650̲1901.indb 14- Por 14 -2019/06/03 14:05

COMO ENROLAR A BRAÇADEIRA DE PULSO

4 Posicione a extremidade

superior da braçadeira afastada
aproximadamente 1 a 2 cm dadobra do seu pulso.

5 Fixe a braçadeira de pulso com ofecho de ganchos e laços.

Português

• Se a braçadeira estiverenrolada muito solta ou muito
apertada, poderão ocorrer
leituras incorretas.

Pode utilizar o seu pulso direito para medir a tensão arterial.
No entanto, a sua tensão arterial pode diferir aproximadamente
10 mmHg dependendo se medir no pulso esquerdo ou direito e,
portanto, idealmente sempre deve medir a sua tensão arterial nomesmo pulso.
Para obter medições precisas• Colocar a braçadeira muito frouxa ou muito apertada pode impedir
uma medição correcta.
• Sente-se erecto enquanto toma a medição.

CH-650̲1901.indb 15- Por 15 -2019/06/03 14:05

SUGESTÕES SOBRE COMO OBTER MEDIÇÕES EXACTASA tensão arterial varia com a postura, a hora do dia e muitos outros
factores. Por essa razão recomenda-se que meça a sua tensão arterial
todos os dias à mesma hora e na mesma postura.

➀ Certifique-se de que sua postura esteja correcta.
Sente-se erecto, respire fundo e relaxe.

➁ Coloque a palma virada

para cima e deixe os dedos
relaxados.

➂ É importante que a

braçadeira fique no
mesmo nível do coração.

Português
CH-650̲1901.indb 16

Coloque uma toalha dobrada ou
algo similar sob o braço.

- Por 16 -2019/06/03 14:05

SUGESTÕES SOBRE COMO OBTER MEDIÇÕES EXACTASPara obter medições correctas

CH-650̲1901.indb 17• Não cerre o punho com força.

• Não mude a posição do medidor.

Português

• Não coloque a braçadeira
demasiadamente frouxa.

• Não mova o corpo nem fale durante a
medição.

- Por 17 -2019/06/03 14:05

MEDIR A TENSÃO ARTERIAL
1 Enrole a braçadeira de pulso à volta do seu pulso esquerdo.
(

Ver página 14 “COMO ENROLAR A BRAÇADEIRA DE PULSO”.)

2 Assegure-se de que a sua postura está correta.
( er página 16 “SUGESTÕES SOBRE COMO OBTER
VMEDIÇÕES EXACTAS”.)

3 Prima o botão “START/STOP” para

iniciar a medição.
• O aparecimento de todos os dígitos novisor é um ecrã inicial que indica que omonitor está a funcionar normalmente.
A marcatambém aparece, masisto não significa que a pilha precisa sersubstituída.

Português
CH-650̲1901.indb 18

é apresentado quando uma pulsação é detectada.

- Por 18 -2019/06/03 14:05

MEDIR A TENSÃO ARTERIAL

4 Os resultados da medição são apresentados.
Pressão sistólica

Pressão diastólica
Indicador depulsação irregular(IHB)

CH-650̲1901.indb 19

Pulsação

O indicador de movimento do corpo é apresentado
na indicação dos resultados da medição se mover amão ou o pulso durante a medição, causando uma
mudança substancial da tensão a ser detectada.
• Se o indicador de movimento do corpo forapresentado, meça a tensão arterial de novo.
Se uma pulsação irregular for detectada durante a
medição, o indicador de pulsação irregular (IHB) seráapresentado após a conclusão da medição.
Uma pulsação irregular é definida como uma pulsação
que varia ±25% desde o meio de intervalos de 5 batidas
do coração durante a medição da pressão arterial.
• A medição pode não ser correcta se a sua
pulsação flutuar muito durante a medição.
Se o indicador de pulsação irregular (IHB) for
apresentado, meça a tensão arterial de novo.

Precaução

Português

Indicador de movimento do corpo

Se o indicador de pulsação irregular (IHB)
aparecer com frequência, é aconselhável
consultar um médico sobre a sua condição
de saúde.

- Por 19 -2019/06/03 14:05

MEDIR A TENSÃO ARTERIAL

5 Terminar a medição.
• Prima o botão “START/STOP” para desligara o aparelho.

FUNÇÃO DE DESLIGAR AUTOMÁTICO
• Depois de efectuar a medição da tensão arterial, premir o botão
“START/STOP” desligará o aparelho. Contudo, se não premir obotão, o aparelho desligar-se-á automaticamente após 3 minutos.

ACEDER AOS DADOS GUARDADOS NA MEMÓRIA
60 resultados de medição podem ser armazenados na memória. O
valor médio é automaticamente calculado para ajudá-lo a administrar asua saúde diária.

COMO ACEDER AOS DADOS
Português
CH-650̲1901.indb 20

1 Prima o botão “MEMORY”.

• “3A”a indicar o valor médio aparece na secção esquerda do visor.
De seguida, o valor médio, com base nas três últimas medições,é apresentado. (“3A” e o valor médio não aparecem se exixtirem
menos que duas medições na memória.)- Por 20 -2019/06/03 14:05

ACEDER AOS DADOS GUARDADOS NA MEMÓRIA

2 Cada vez que preme o botão, os dados de medição são

apresentados em ordem, a partir do mais recente para o mais antigo.
O número dos dadosé apresentado antes
dos resultados demedição.

• O valor guardado na memória é numerado pela ordem das medições. Por
exemplo, se na memória estiverem memorizadas 60 medições, a medição
número 1 será a última medição realizada (ou seja, a medição mais recente).
Número dos dados: 1, 2, 3, ... ... 59, 60ÇÇDados mais recentes
Dados mais antigos

COMO APAGAR OS DADOS
1 Prima o botão “MEMORY”.
2 Enquanto preme o botão “MEMORY”, prima o botão “START/

Português

Os resultados de medição são apresentados após o número dos dados.

STOP” durante mais de 3 segundos.
Todos os dados armazenados serão apagados.

CH-650̲1901.indb 21

Prima
durante 3
segundos

Apagados

É apresentada
uma mensagema dizer que todos
os dados foram
apagados damemória.

- Por 21 -2019/06/03 14:05

INDICAÇÕES DE ERROMARCA VISUALIZADA

Português
CH-650̲1901.indb 22

CONDiÇÃO/CAUSAAÇÃO CORRECTIVA

Aparece quando a
braçadeira de pulso éenrolada ficando muito
solta.

Volte a enrolar a braçadeirade pulso ajustando-a
confortavelmente. Tomeoutra leitura, assegurandose de que se mantém
imóvel. (Ver página 14“COMO ENROLAR A
BRAÇADEIRA DE PULSO”.)

Aparece quando o seu
pulso se moveu durante
a medição; ou Aparecequando a braçadeiranão está corretamente
enrolada.

Volte a enrolar corretamentea braçadeira de pulso.
Tome outra leitura,assegurando-se que
permanece perfeitamente
imóvel. (Ver página 14“COMO ENROLAR A
BRAÇADEIRA DE PULSO”.)

Aparece quando as
pilhas se encontram
gastas.

Substitua as duas pilhas
alcalinas por novas
Ver página 13(“RETIRAR AS PILHAS”.)

Aparece quando apulsação fica fora dos
limites de medição
(inferior a 39 batimentos/
min. ou superior a 181
batimentos/min.)Volte a enrolar
corretamente a braçadeirade pulso. Respire fundopara relaxar, e meçaVernovamente. (página 16 “SUGESTÕES
SOBRE COMO OBTERMEDIÇÕES EXACTAS”.)

A unidade não funciona
adequadamente.

Contate a loja onde esta
foi comprada ou o seu
centro de assistência
Ver página
local. (25 “ANTES DE SOLICITARREPARAÇÕES OU
INSPECÇÃO”.)- Por 22 -2019/06/03 14:05

SOBRE A TENSÃO ARTERIAL
O que é a tensão arterial?

O coração é uma bomba que faz circular o sangue através do corpo. O sangue ébombeado do coração para as artérias a uma pressão constante. Esta pressão é
designada de pressão sanguínea arterial e, em termos gerais, representa a tensão
arterial. A tensão arterial é indicada por vários tipos de pressões, tais como pressão
sistólica que é a pressão que ocorre quando o coração bombeia o sangue e pressão
diastólica que é a pressão que ocorre quando o sangue regressa ao coração.
(Sistólica)(Diastólica)
ArtériaÁtrioesquerdo
Ventrículo
esquerdo

ArtériaÁtrioesquerdo

A parede da
artéria expande.

Ventrículo
esquerdo

Contracção

A parede da artéria volta
ao estado normal.
Dilatação

A tensão arterial varia segundo a idade, sexo e muitos outros factores. A tensão
arterial de uma pessoa pode ser afectada pelo biorritmo durante o dia e por
postura, actividades físicas, actividades mentais, nível de stress e até mesmo pela
temperatura atmosférica. Diz-se que a tensão arterial de uma pessoa saudável varia.

Valor da pressão sanguínea (mmHg)[Exemplo das variações da pressão arterial no espaço de um dia]
Pressão arterial
sistólica

160

Português

A tensão arterial está sempre a mudar.

140
120
100
80
60
40

A dormirPressão arterial
diastólica

12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11

Hora do dia(Os valores de pressão arterial medidos em intervalos de cinco minutos ao longo do dia)l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329–44.

CH-650̲1901.indb 23- Por 23 -2019/06/03 14:05

PERGUNTAS E RESPOSTAS SOBRE A TENSÃO ARTERIALP Porque que é que a tensão arterial medida em casa é diferente da
medida por médicos ou em hospitais?

R A tensão arterial pode variar com o exercício, com a temperatura

ambiente ou com estados psíquicos. Quando um médico ou enfermeiramede a sua tensão arterial, o valor da tensão arterial é provavelmente 10
a 20 mmHg superior ao normal devido à preocupação e/ou nervosismo.
Saber o que afecta a sua tensão arterial ajuda a monitorizar a sua saúde.

P Por que razão a tensão arterial é diferente em cada medição?
R A pressão sanguínea é ajustada pelo funcionamento do sistema nervoso
automático. A tensão arterial é diferente a cada batimento cardíaco.
Provavelmente pensa que a tensão arterial é constante, mas na verdadea tensão arterial não é constante e obterá sempre valores diferentes serealizar medições umas a seguir às outras. A tensão arterial muda com
a hora do dia, a estação do ano e a temperatura atmosférica. Para alémdisso, a tensão arterial é facilmente influenciada pelo stress mental ou
por factores emocionais. A tensão arterial tende a aumentar quando seestá nervoso e a diminuir quando se está descontraído.

Português
CH-650̲1901.indb 24

P Qual o benefício de medir a tensão arterial em casa?
R Medir a tensão arterial em casa permite obter dados fiáveis já que amedição pode ser feita quando está calmo e descontraído. Os médicos
dão muita importância às medições da tensão arterial feitas em casa.
Recomenda-se que meça a sua tensão arterial todos os dias à mesma
hora e que registe as variações da tensão arterial.

Para a interpretação dos valores da tensão arterial e parao tratamento adequado, consulte sempre o seu médico.

- Por 24 -2019/06/03 14:05

ANTES DE SOLICITAR REPARAÇÕES OU INSPECÇÃO
Assegure-se de que verifica o seguinte antes de enviar o seu medidor
da tensão arterial para reparação ou inspecção.
Anomalia

Verificações

Acção correctiva

O visor continua
vazio quando
prime o botão
“START/STOP”.

Verifique se as pilhasestão gastas.

Substitua todas as pilhas
por pilhas novas.

Verifique se aspilhas estão bem
colocadas (comos sinais de e
virados para o ladocorrecto).

Coloque as pilhas na
direcção correcta.

Verifique se o ícone
aparece.

Coloque a braçadeira deforma adequada.

Verifique se a
braçadeira está bem
colocada.
Permaneceu calmo
durante a medição?

Meça a tensão de novo,
assegurando-se de que
permanece quieto.

Note que pode não ser possível medir a tensão
arterial em pessoas com pulso muito fraco ou
arritmia cardíaca (pulsação irregular).
Os resultados da medição são muito
mais elevados ou muito mais baixos
do que os das medições realizadas no
hospital.
Outros fenómenos

CH-650̲1901.indb 25

Português

Não seconsegue medira tensão.

Consulte a página 24.

Retire as pilhas do aparelho
e substitua-as por pilhas
novas.

- Por 25 -2019/06/03 14:05

ESPECIFICAÇÕES
Modelo número
Sistema de medição
Visor
Perímetro do pulso
Amplitude de Pressão
medição
Pulsação
Pressão
Nível dePulsação
precisão
Insuflação
Tensão nominal
Saída de arTensão de alimentação
Duração das pilhas

Português
CH-650̲1901.indb 26

Condições de Temperatura
funcionamento Humidade
Temperatura
Condições dearmazenamento HumidadeFunção da Memória
Protecção contra choqueeléctrico
Grau de protecção:Modo de operação
Protecção contra o ingressode águaDimensões
Peso
Vida útil
Acessórios

CH-650
Método oscilométrico
Tipo de visor digital
13,5 a 21,5 cm
0 a 280 mmHg
40 a 180 pulsações/minuto±3 mmHg±5% da leitura
Insuflação automática por bomba interna
CC de 3 V(
: Corrente contínua)Válvula electromagnética de saída rápida
de ar
2 pilhas AAA (LR03)(180 mmHg, uma
Approx. 400 vezes
vez por dia, 22°C)
10°C a 40°C
15% a 85% RH-20°C a 60°C
10% a 95%RH
Até 60 medições
Unidade de alimentação internaParte aplicada tipo BF
Funcionamento contínuo
IPX065(L) x 74(A) x 29(P) mm
Aprox. 103 g (excluindo pilhas)
5 anosConjunto composto por 2 pilhas AAA(LR03) para o medidor, manual de
instruções, estojo de arrumação

* A peça aplicada para este dispositivo é Braçadeira.
* Um intervalo na pressão barométrica de 700 hPa a 1060 hPa* A precisão total do sistema deste dispositivo está em conformidade com orequisito do item 7.9 da norma EN1060-3.
EN1060-3: Esfignomanómetro não invasivo Parte 3 (Norma Europeia)- Por 26 -2019/06/03 14:05

CH-650̲1901.indb 27

MEMO

2019/06/03 14:05

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 28

2019/06/03 14:05

CH-650̲1901.indb 1

전자 혈압계
사용설명서

한국어CH-650한국어

1901

2019/06/03 14:05

목차
3 - 일반적인 사항
5 - 사용 및 관리상의 주의사항10 - 제품 구성 및 명칭
1２ - 배터리 삽입방법1３ - 배터리 교체방법1４ - 손목 가압대 착용 방법
1６ - 정확한 측정 결과를 얻는 방법
18 - 혈압 측정 방법
20 - 저장된 데이터를 불러내는 방법
22 - 화면에 나타나는 표시들
23 - 혈압에 대해서
24 - 혈압에 관한 질문과 답변
25 - 고장수리 맡기기 전에
26 - SPECIFICATIONS
27 - 품질보증서
26 - 국문표시사항

한국어
CH-650̲1901.indb 2다음 부품이 모두 들어 있는지 확인하십시오.

혈압계 본체

보관함（모니터용）
2AAA 사이즈
배터리

사용설명서

- Kor 2 -2019/06/03 14:05

CH-650̲1901.indb 3한국어일반적인 사항혈압 측정 시 주의점
1. 가압대(Cuff)에 공기가 빨리 빠지는 경우, 전원버튼을 누르십시오.
2. 부착된 가압대(Cuff)에 알맞은 팔목 둘레는 13.5~21.5cm
입니다.
3. 가압대(Cuff)를 부착할 때, 혈압계의 모니터가 팔목의 안쪽에 위치해야 합니다.
4. 본 기기는 자가측정용으로서 절대 치료용으로 사용될 수없습니다.
5. 자가측정을 통하여 임의로 담당 의사가 규정한 약물 투약량을 바꾸시면 안됩니다.
6. 혈압을 올바르게 측정하기 위해서는 측정하기 10분~15분
전에 소변을 보시고 어느 정도의 안정을 취하셔야 합니다.
7. 혈압 측정 전의 운동, 음식물 섭취, 음주, 흡연 등은 결과에 영향을 끼칠 수 있습니다.
8. 혈압은 수시로 변하므로 혈압측정은 같은 시간대에 매일이루어져야 합니다.
9. 절대로 한 개의 측정치를 가지고 불안해 하시거나 안도하지 마시고, 혈압의 변화를 기록하십시오. 누적된 수치만이 건강상태를 바르게 나타내 줍니다.
10. 정신적인 스트레스가 혈압 상승에 원인이 될 수 있습니다.
올바른 혈압 측정 방법
1. 혈압 측정 전 심호흡을 5~6번 하여 긴장을 풉니다. 긴장한상태에서 측정하면 제대로 된 수치를 얻을 수 없습니다.
2. 짜증나거나 화가 난 상태, 수면부족, 변비, 운동 직후나식후일 경우에는 혈압이 높아집니다.
3. 가압대(Cuff)를 손목에 편안하게 착용하시오.（14페이지 참조）4. 음주나 목욕 후에 혈압을 측정하지 마시오.
5. 소변을 보셔야 할 필요가 있다면, 측정하기 전에 소변을 보시오.
6. 실내 온도가 20℃일 때가 가장 적당하며, 10℃이하나
40℃이상일 경우 측정하지 마시오.
7. 커피나 차, 흡연 후에 바로 측정하지 마시오.
8. 안정되고 편안할 때 혈압을 측정하시오. 가압대(Cuff)를
심장 높이에 맞추고 팔을 움직이거나 얘기하지 마시오.
9. 가압대(Cuff)를 착용한지 몇 분이 지나거나 그 이상 되면, 측정하지 마십시오. 높은 혈압수치가 나오게 되며,유효한 측정값이 나오지 않습니다.
- Kor 3 -2019/06/03 14:05

일반적인 사항10. 신체 흔들림 표시 또는 불규칙적인 심장박동(IHB) 표시가 나타나면 팔의 힘을 빼고 혈압을 다시 측정하십시오.
불규칙적인 심장박동(IHB) 표시가 자주 나타날 경우 의사에게 건강 상태를 검진받는 것이 좋습니다. （19
페이지 참조）11. 다음에 나타난 표는 표준적인 혈압의 건강상태를 WHO기준에 따라 나타낸 것입니다.
수축기（mmHg）
180

3등급 고혈압(고도)2등급 고혈압(중도)160주의:그래프는 정확하지는 않
습니다. 하지만 대강의 혈압측정방법을 이해할 수
있습니다. 기기는 어른을 위한 용도로 사용해야합니다.

1등급 고혈압(경도)
140
130
120

높음-정상
정상
최적
80 85 90

100 110 이완기（mmHg）

혈압 수치의 정의 및 분류한국어CH-650̲1901.indb 4

12. 사용 목적• 본 장치는 오실로메트릭법을 사용한 비침습 혈압계입니다. 본 장치는 수축기 혈압(SYS), 이완기 혈압(DIA)및 맥박을 자동으로 측정할 수 있습니다.
• 본 장치는 의사의 지시 하에 가정에서 개인의 건강 상태를검사하는 데 사용되며, 진단 장치로 제공되지 않습니다.
• 본 장치는 신생아 또는 유아에게 사용하거나 자동 순환측정을 수행하기 위한 용도로 제작되지 않았습니다.
혈압을 매일 같은 시간대에 측정하십시오.
* 혈압은 매시간 마다 변합니다. 오랜 시간 동안 여러 번 측정하는 것보다 하루에 한번 측정하는 것이 더욱 정확한 측정입
니다. 이러한 이유로 매일 같은 시간대에 당신의 혈압을 측정
하십시오. 어디서나 당신의 이상적인 시간에 측정하십시오.
- Kor 4 -2019/06/03 14:05

CH-650̲1901.indb 5한국어사용 및 관리상의 주의사항사용상의 주의사항
1. 심장병을 가지고 있거나 높은 혈압이나 다른 순환계 질환은 측정계를 사용하기 전에 의사를 만나보시오.
2. 사용 중 가압대(Cuff)의 느낌이 이상하거나 불규칙하다
면, 혈압계의 “START/STOP” 버튼을 누르신 뒤 가압을 줄이십시오. 가압대(Cuff)를 푸시거나 혹은 공기 연결을 떼내신 후 판매점에 문의하시오.
3. 측정값이 이상하거나 측정값이 좋지 못하면, 사용을 중지하고 의사와 상담하시오.
4. 맥박이 약하거나 부정맥이 있는 사람에겐 혈압측정이 불가능 할 수 있습니다.
5. 반복적인 혈압계 사용은 울혈이나 부종이 나타날 수 있
습니다.
6. 계속적인 혈압 측정 시 정확한 측정값이 나오지 않습니
다. 1분 이상의 간격을 두시고 다시 측정 하시오.
7. 순환계 질병 중 팔에 관한 질병을 가지고 있다면, 기기를사용하기 전에 의사와 상담하시오. 건강에 위험할 수 있
습니다.
8. 충분하게 혈액이 순환하지 않거나, 불규칙적인 심박수를가지고 있다면, 기기 사용을 하지 마시고 의사와 상담한후에 사용하시오.
9. 상처 난 팔 주위에 가압대(Cuff)를 사용하지 마시오.
10. 정맥 주사를 맞은 손목 주위에나 혈액 수혈 뒤에 가압대(Cuff)를 사용하지 마십시오. 부상이나 심각한 사고가 날수 있습니다.
11. 가연성 기체가 있는 부근에서 혈압측정기를 사용하지 마
십시오. 산소와 반응하거나 공기 중에 점화되어 폭발할수 있습니다.
12. 병원의 고압실이나 산소공급용 텐트와 같은 환경에서 기기를 작동하지 마시오. 산소가 점화되어 폭발할 수 있습니다.
- Kor 5 -2019/06/03 14:05

사용 및 관리상의 주의사항13. 기기 주변에서 휴대 전화를 사용하지 마시오. 오작동 할수 있습니다.
14. 심장박동 조절장치를 사용하고 계시다면, 기기 사용에대해 전문의와 상담하시오.
15. 이 기기는 혈압 측정용으로만 사용하십시오. 다른 목적으로 절대 사용하지 마시오.
16. 허가 받은 부품과 액세서리만 사용하십시오. 허가 받지않은 부품과 액세서리를 사용할 경우 장치가 손상될 수있습니다.
17. 유아에게 사용하지 마시오.
18. 부정맥 환자나 심방 혹은 심실 조숙증, 심방 섬유증 환자는 혈압측정이 불가능 할 수 있습니다.

한국어
CH-650̲1901.indb 6지속적으로 주의해야 할 사항
1. 직사광선이 직접적으로 영향을 주는 곳에 기기를 놓지
마시오. 고온(60℃ 이상)이나 저온(–20℃이하)에서 보관하지 마시오. 상대적으로 높은 습도(95%이상) 혹은 먼지에 놓여있게 하지 마시오.혈압계는 어린이, 애완동물및/또는 유해동물이 없는 장소에 보관하십시오.
2. 심한 충격을 주거나 떨어뜨리지 마시오.
3. 긴 기간 동안 기기를 사용하지 않는 다면, 배터리를 빼놓으시오.
4. 기기를 분해하려고 하지 마시오.
5. 가압대(Cuff)를 구부리거나 공기관을 과도하게 구부리지마시오.
6. 기기가 더럽다면, 촉촉하고 살균효과가 있는 것으로 닦으시오.
7. 기기를 희석제나 벤젠 혹은 기기에 손상을 줄 수 있는 것을 이용하지 마시오.
8. 가압대(Cuff)를 닦기 위해선 촉촉한 것을 이용하여 닦으
시오. 거친 것으로 문지르지 마시오. 공기가 세어 나갈수 있습니다. 공기관에 물이 들어가지 않도록 하십시오.
- Kor 6 -2019/06/03 14:05

사용 및 관리상의 주의사항경고 본 장비의 개조는 허용되지 않습니다 .
기호 설명

CH-650̲1901.indb 7: CE 표시는 EU 회원국의 시장 조사관에게 정보를제공하는 데 사용됩니다.
: BF형 장착부: 습기주의: 경고: 사용하기 전에 사용설명서 참조: WEEE 지침 준수 기기: 제조업체한국어경고본 장치는 EMC 의 EN60601-1-2 에따라 테스트 및 승인을 받았습니다 . 이는본 장치가 어떤 식으로든 전자기 간섭의영향을 받지 않음을 보장하지 않습니다 .
전자기가 높은 환경에서는 장치를 사용하지마십시오 .

- Kor 7 -2019/06/03 14:05

사용 및 관리상의 주의사항유지보수의 성격 및 빈도 :
본 제품은 장기간에 걸쳐 사용할 수 있도록 설계되었으나 , 올바른 기능과성능을 보장하기 위해서는 일반적으로 2 년마다 검사하는 것이 좋습니다 .

자연 환경 보호 :사용 기간이 다 된 배터리와 제품을 폐기할 때는 국가 및 / 또는지역의 재활용 규정을 준수하여 자연 환경을 보호해 주십시오 .

WEEE 마크본 제품을 폐기하려면 일반 가정용 폐기물과 섞지마십시오 . 유럽 연합 내에서만 유효한 WEEE 지침 ( 지침2012/19/EU) 의 규정에 따라 폐전자제품에 적용되는별도의 수거 시스템이 있습니다 .

유럽 연합 이외 국가의 폐기 정보 .
다 쓴 배터리를 폐기하려면 현지 지방자치단체 또는 대리점에 올바른폐기 방법을 문의하십시오 .

배터리 기호 참고사항 .
이 기호는 화학 기호와 함께 사용될 수 있습니다 . 이 경우 관련화학물을 위한 지침에 규정된 요건을 준수합니다 .

유럽 표준 참조 :혈압 측정 장치는 EN60601-1, EN1060-1, EN1060-3, EN1060-4 규정에준거합니다 .

보정 :한국어CH-650̲1901.indb 8이 혈압 측정 장치는 올바른 기능과 정확성을 보장하기 위해 일반
적으로 2년마다 모니터를 검사하는 것이 좋습니다. 대리점에 문의하십시오.

- Kor 8 -2019/06/03 14:05

사용 및 관리상의 주의사항

이름 :
주소 :

공장
이름 :
주소 :

CITIZEN SYSTEMS JAPAN CO., LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi, Tokyo 188-8511, Japan
CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
Building 6, 399 Jinxing Road, Jianghai District, Jiangmen,
Guangdong, China

유럽 총괄 대리점 :
이름 :
주소 :

EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague,
The Netherlands• 기술 데이터는 유럽 총괄 대리점에 문의하십시오 .

CH-650̲1901.indb 9

한국어

중국산

- Kor 9 -2019/06/03 14:05

제품 구성 및 명칭
부품 이름
<본체>

화면

메모리 버튼
START/STOP버튼 대기배터리 커버
한국어
CH-650̲1901.indb 10손목 가압대(cuff)- Kor 10 -2019/06/03 14:05

제품 구성 및 명칭<액정화면>

확장기혈압
표시

수축기혈압
표시

/STOP
대기

불규칙적인
심장박동 표시신체 움직임
표시

CH-650̲1901.indb 11건전지양 표시배터리 잔량이 낮아지면 표시됩니다. （13페이지 참조）

한국어

맥박 표시

- Kor 11 -2019/06/03 14:05

배터리 삽입방법
（혈압계를 사용하기 전에 기기와 함께 제공된 배터리를배터리 삽입구에 장착해야 합니다 .)1 배터리 커버를 여시오.

배터리 커버배터리 커버를 누르면서커버를 화살표 방향으로밀어 제거합니다.

2 배터리 안에 삽입하십시오.
케이스의 , 표시에 맞추어 배터리를 삽입합니다.
* 부분을 먼저 끼우세요한국어
CH-650̲1901.indb 12배터리의 표시 부분을 먼저 끼웁니다.

- Kor 12 -2019/06/03 14:05

배터리 삽입방법

3 배터리 커버를 닫으시오.
커버를 화살표 방향으로딸깍 소리가 날 때까지 밀어서 닫습니다.

• 액정화면에표시가 나타나면, 배터리를 2개 모두 교체하십시오.
• 알카라인 배터리를 사용해야 합니다.
• 충전용 배터리는 사용해서는 안됩니다.
• 알카라인(LR03) 배터리와 망간(R03) 배터리를 함께 사용해서는 안됩니다.
• 제품에 포함된 배터리는 오래 사용하실 수 없습니다.
• 환경을 위하여 다 사용된 배터리를 규정에 맞게 처리하시기 바랍니다.

CH-650̲1901.indb 13한국어배터리 교체방법

- Kor 13 -2019/06/03 14:05

손목 가압대 착용 방법손목 가압대를 손목 주위에 올바르게 감으십시오. 가압대를잘못 감으면 측정 오류가 발생할 수 있습니다.

1 손목 가압대를 엽니다.
2 기기의 화면이 손목 안쪽에위치하도록 하십시오.

• 맨살 피부위에 손목 가압대(Cuff)를 감으시오.
• 손목둘레는 13.5~21.5cm가
적당합니다.

3 손목 가압대(Cuff)를 손목에 놓고 너무 느슨하거나 너무 조이지 않도록착용하십시오.

한국어
CH-650̲1901.indb 14- Kor 14 -2019/06/03 14:05

측정방법

4 가압대(Cuff)의 가장자리가손목 주름 부분에서 1~2cm
위에 위치하도록 하시오.

5 손목 가압대(Cuff)를 안전하게 고정시키시오오른쪽 손목을 사용하여 혈압을 측정할 수도 있습니다.

한국어• 만약 손목 가압대(Cuff)를 너무 느슨하거나 너무 조이게 착용하면 에러가 나거나측정값이 정확하지 않을 수 있습니다.

오른쪽 손목으로 측정한 값과 왼쪽 손목으로 측정한 값이
10mmHg 차이가 날 수 있으므로 매회 같은 손목으로 측정하시오.

정확한 측정 결과를 얻으려면• 가압대를 너무 느슨하거나 너무 조이게 감으면 측정이 정확하지 않을 수 있습니다.
• 측정 도중에는 등을 곧게 펴서 앉습니다.

CH-650̲1901.indb 15- Kor 15 -2019/06/03 14:05

정확한 측정 결과를 얻는 방법측정하는 자세에 따라서 측정값이 달라질 수 있으므로 매일같은 시간대에 같은 자세로 측정하십시오.

➀ 똑바로 앉아 심호흡을 하고 편한 자세를 취하시오.

➁ 손바닥을 위로 향하게 하고손가락의 힘을 빼시오.

➂ 가압대(Cuff)를 심장과같은 위치에 놓고 측정
하시오.

한국어
CH-650̲1901.indb 16

접힌 수건이나 그와 유사한것을 팔목 아래에 놓고 측정합니다.

- Kor 16 -2019/06/03 14:05

정확한 측정 결과를 얻는 방법올바른 측정 결과를 얻으려면CH-650̲1901.indb 17

주먹을 꽉 쥐지 마시오.

팔목 가압대(Cuff)를 너무 느슨하게 감지 마시오.

측정시 몸을 움직이지 마시고 말을 하지 마시오.

한국어측정기를 움직이지 마시오.

- Kor 17 -2019/06/03 14:05

혈압 측정 방법
1 손목 가압대(Cuff)를 왼쪽 손목에 감으시오.
（14페이지 “손목 가압대 착용 방법”참조）

2 자세가 올바른지 확인하시오.
（16페이지 “정확한 측정결과를 얻는 방법”참조）

3 START/STOP 버튼을 누르시오.
•한국어
CH-650̲1901.indb 18

표시된 모든 숫자는 모니터가 정상적으로 작동하고 있음을 나타내는 초기 표시값입니다.
표시도나타나지만 배터리를 교체해야 한다는 의미는 아닙니다.

는 맥박을 감지했을 때 표시되어 집니다.

- Kor 18 -2019/06/03 14:05

혈압 측정 방법

4 측정결과가 화면에 표시됩니다.
확장기혈압
표시
수축기혈압
표시

불규칙적인 심장박동 표시

CH-650̲1901.indb 19맥박신체 흔들림 표시측정 도중 불규칙적인 심장 박동이 감지되면 측정이 완료된 후 불규칙적인 심장박동(IHB) 표시가 나타납니다.
불규칙적인 심장 박동이란 혈압 측정 도중 5회의심장 박동 간격의 중간 값에서 ±25% 범위까지 변동하는 심장 박동을 의미합니다.
• 측정 도중 심장 박동이 크게 변동하는 경우 측정이 정확하지 않을 수 있습니다. 불규칙적인 심장박동(IHB) 표시가 나타나면 혈압을 다시 측정
하십시오.

주의

한국어

신체 움직임 감지 표시는 당신의 손이나 팔이 흔들릴 경우 기기화면에 나타나게 되며, 상당한 혈압 변화가 나타납니다. 신체 움직임 표시가 나타나게 되면 혈압을 다시 측정하시오.

만약 자주 불규칙적인 심장 박동이감지되면 , 전문의와 상담을 하셔야합니다 .
- Kor 19 -2019/06/03 14:05

혈압 측정 방법

5 측정 종료

• START/STOP 버튼을 눌러 화면을 종료합니다.

자동 전원 종료• 혈압 종료 후, START/STOP 버튼을 누르지 않으면 3분후에 자동 종료 됩니다.

저장된 데이터를 불러내는 방법60회의 측정 결과를 메모리에 저장할 수 있습니다.
평균값이 자동으로 계산되므로 매일 건강을 관리할 수 있습니다.

데이터 불러내기
1 메모리 버튼을 누르시오.

한국어
CH-650̲1901.indb 20• “3A”는 최근 측정한 3번의 평균값을 표시합니다. 만약저장되어진 메모리가 2개 또는 2개 이하인 경우“3A”와
평균수치는 나타나지 않습니다.

- Kor 20 -2019/06/03 14:05

저장된 데이터를 불러내는 방법

2 스위치 버튼을 누를 때마다 가장 최근의 측정값이표시되어 집니다.

측정값이 나타나기 전에 숫자가
표시되어 진다.

측정 결과는 데이터 번호가 나타난 후에 표시됩니다.
•	데이터 번호가 높을수록 오래된 측정값입니다. 예를 들어60개의 값을 저장했을 경우 60은 가장 처음 즉 가장 오래된 데이터이고 1은 가장 최근의 데이터입니다.
데이터 번호: 1, 2, 3, ... ... 59, 60ÇÇ최근 데이터최초 데이터

1 메모리 버튼을 누르시오.
2 메모리 버튼과 START/STOP 버튼을 동시에 3초

CH-650̲1901.indb 21

한국어저장된 데이터 지우는 방법이상 누르시오.

모든 데이터는 지워집니다.
3초 동안
누름

삭제됨

이 표시가 나타나면 모든 데이터가 지워진 것입니다.

- Kor 21 -2019/06/03 14:05

화면에 나타나는 표시들
화면 표시

상태/진행

행동 방안

손목 가압대(Cuff) 손목 가압대(Cuff)가 너무 느슨하게 를 다시 감고 다시감겨졌을 때.
측정 하시오.
（14페이지“손목 가압대 착용 방법”참조）측정하는 동안 손 손목 가압대(Cuff)목이 움직이거나를 올바르게 다
가압대(Cuff)가 올 시 착용하고 측정바르게 감겨지지
하시오.（
14
않았을때.
페이지 “손목 가압대 착용 방법”참조）배터리의 수명이
2개의 배터리를
다했을때.
교체 하시오.
（13페이지“배터리 교체방법”참조）한국어CH-650̲1901.indb 22

맥박의 측정범위
심호흡을 한 후,를 벗어났을때다시 측정하시
(39이하 181이상) 오.（16페이지 “정확한 측정결과를 얻는 방법”참조）기기내의 결함기기를 구입한 곳일 때.
이나 서비스센터에 문의하시오.
（25페이지
“고장수리 맡기기 전에”참조）- Kor 22 -2019/06/03 14:05

혈압에 대해서
혈압이란?
심장은 우리 몸의 피를 순환시키는 펌프 역할을 합니다.
피가 심장에서 동맥으로 나가는 과정에서의 규칙적인 압력이 우리가 말하는 혈
압입니다. 혈압은 심장에서 혈액을 내보낼 때 발생하는 수축기 혈압과 혈액이 심장으로 돌아올 때 발생하는 이완기 혈압을 포함한 여러 종류의 혈압으로 표시
됩니다.
(수축기)
(이완기)
동맥
동맥
좌심방

좌심실

좌심방

동맥 벽이 확
장됩니다.
수축

좌심실

동맥 벽이 정상 상태로돌아갑니다.
팽창혈압은 항상 바뀝니다.
혈압은 연령, 성별 및 기타 다양한 요인에 따라 달라집니다. 혈압은 하루 동안의 바이오리듬, 자세, 신체 활동, 정신 활동, 스트레스 수준, 심지어 공기 온도에도 영향을 받을 수 있습니다. 건강한 사람의 혈압은 일반적으로 하루 동안 계속 변합니다.

[아래의 차트는 하루 동안의 혈압의 변화를 보여줍니다.]수축기 혈압

160

한국어혈압값(mmHg)

140
120
100
80
60

이완기 혈압

40

수면
12 13 14 15 16 17 18 19 20 21 22 23 24 1 2 3 4 5 6 7 8 9 10 11시각(혈압값은 하루 동안 5분 간격으로 측정됨)l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329-44.

CH-650̲1901.indb 23- Kor 23 -2019/06/03 14:05

혈압에 관한 질문과 답변Q 왜 집에서 측정한 혈압과 병원에서 측정한 혈압은 다를까요?

A 혈압은 보통 심리 상태에 쉽게 영향을 받습니다. 의사나간호사가 혈압을 잴 경우 걱정이나 불안감 때문에 혈압이 약간 높게 나오는 경우가 있습니다. (평소보다 최고
10~20mmHg가 더 높게 나올 수도 있습니다.) 반면에 집에서는 안정된 상태이기 때문에 정상치 혈압이 나올 확률이 높습니다.

Q 왜 모든 혈압 수치는 항상 다를까요?
A 우리 몸의 혈압은 자율신경의 기능에 의해 조정 됩니다.

혈압은 시간대, 계절, 온도에 민감하여 특히 정신적 스트레스나 감정의 상태에 따라 쉽게 영향을 받습니다.
따라서 긴장을 하면 혈압이 증가하고 안정되면 감소하게 됩니다.

Q 집에서 혈압을 측정하는 의미는?
A 집은 안정되고 편한 환경이기 때문에 혈압을 측정하기한국어
CH-650̲1901.indb 24

좋습니다. 단 한번의 측정치로 건강 상태를 판단하기보다 매일 집에서 같은 시간대 에 혈압을 재면서 가족 주치의와 상담하는 것이 좋습니다.

항상 의사에게 혈압 수치에 대한 해석을문의하고 적절한 치료법을 결정하십시오 .

- Kor 24 -2019/06/03 14:05

고장수리 맡기기 전에수리를 맡기기 전에 미리 확인하십시오.
증상

확인할 점

조치방법

START/STOP 버튼을눌러도 화면에 아무 표시가 없을 때.

건전지 잔량 확인.

건전지를 새 건전지로교체하시오.

건전지의 , 극이 제대로 맞추어져 있는지 확인.

건전지를 정해진 극에맞추어 주시오.

표시가 나타나는지 확인하시오.

가압대를 확실히 부착하시오.

측정값이 안나올 때.

가압대(Cuff)가
제대로 부착되어 있는지 확인하시오.
측정하는 동안 움직이지 않았는지
확인하시오.

움직이지 말고 진정한상태에서 다시 측정하시오.

당신의 혈압이 비정상적으로 높거나 병원에서 측정한 결과보다 낮게 나올 경우.

기기에서 건전지를 빼낸 후에 새 건전지로 교체해 보시오.

위와 다른 현상CH-650̲1901.indb 2524`페이지 참조한국어혈압이 지나치게 약하거나 비정상적인 맥박을 가진 경우 측정할 수 없을 수도 있음.

- Kor 25 -2019/06/03 14:05

SPECIFICATIONS
Model Number
Measurement system
Display
Wrist circumference range
Pressure
Measurement
range
Pulse
Pressure
Accuracy
Pulse
Inflation
Rated voltage
ExhaustPower supply
Battery duration

한국어
CH-650̲1901.indb 26

Temperature
Operating
Humidity
conditions
Temperature
Storageconditions
Humidity
Memory Function
Electric shock Protection
Degree of Protection
Mode of operation
Protection against ingress ofwater
Dimensions
WeightService Life
Accessories

CH-650
Oscillometric method
Digital display type13.5 to 21.5 cm
0 to 280 mmHg
40 to 180 pulse/min±3 mmHg±5% of reading
Automatic inflation by internal pump(
: direct current)
3V DC
Electromagnetic quick exhaust valve
2 AAA batteries (LR03)(180 mmHg,
Approx. 400 times
once/day, 22°C)
10°
C to 40°C
15% to 85% RH-20°C to 60°
C
10% to 95%RH
Up to 60 measurements
Internal power unit
Type BF applied part
Continuous operation
IPX065(W) x 74(H) x 29(D) mm
Approx. 103 g w/o batteries
5 years
Set includes 2 AAA batteries (LR03)
for the monitor, Instruction Manual,
Storage case

* Applied part for this device is Cuff.
* A range in barometoric pressure 700 hPa to 1060 hPa*O verall system accuracy of this device is met with the requirement of
item 7.9 of EN1060-3.
EN1060-3 : Non-invasive sphygmomanometers Part 3 (European
Standard)- Kor 26 -2019/06/03 14:05

품질보증서
1. 본 제품은 엄정한 품질관리 및 검사과정을 거쳐서 만들어진 제품입니다.
2. 소비자의 정상적인 사용상태에서 고장이 발생하였을 경우보증기간 동안은 무상수리를 해드립니다.(품질보증기간:구입일로 부터 1년간)
3. 보증기간 이내라 하더라도 소비자의 과실 또는 천재지변의 경우는 유상 서비스 해드립니다.
4. 수리를 요할 때는 보증서를 꼭 제시해 주십시오.
5. 제품 구매처가 명시되어 있어야 무상 품질보증이 가능합니다.
제품명시티즌혈압계

보증기간

모델명

MODEL CH-650

1년

구매처
판매일자

전화:
년       월      일CH-650̲1901.indb 27

성     명전화:한국어구입자 주소

- Kor 27 -2019/06/03 14:05

국문표시사항제 품 명 자동전자혈압계
형

명

수 입 원

CH-650디아센스코리아㈜ ☎ 02-566-9697
서울 강남구 역삼동 669-5 영신빌딩 4층제 조 원 Citizen Systems Japan Co., Ltd.
허가번호수허111266호

전원

DC 3V
(2*1.5V AAA 배터리)오실로메트릭 방식에 의해 체외에서 혈압을사용목적 측정하는 기구“본 기기는 의료기기 임”
103g ( 배터리, 커프 포
함)

포장단위

1대

제조번호 표시사항참조

제조년월일표시사항참조

보호형식 BF

성능, 사용방법 및 주의사항 :사용설명서 참조중
한국어CH-650̲1901.indb 28

량- Kor 28 -2019/06/03 14:05

CH-650̲1901.indb 29

MEMO

2019/06/03 14:05

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901.indb 30

2019/06/03 14:05

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.
• Design and specifications are subject to change without notice.

CH-650̲1901̲Ara.indd 28

2019/06/03 13:49

CH-650̲1901̲Ara.indd 27

MEMO

2019/06/03 13:49

‫المواصفات‬‫رقم الطراز‬‫نظام القياس‬‫الشاشة‬‫نطاق محيط المعصم‬‫نطاق القياس‬‫الدقة‬‫النفخ‬‫الفولتية المقدرة‬‫التفريغ‬‫مصدر اإلمداد بالطاقة‬‫الضغط‬‫النبض‬‫الضغط‬‫النبض‬‫مدة بقاء البطاريات‬‫ظروف التشغيل‬‫ظروف التخزين‬‫درجة الحرارة‬‫الرطوبة‬‫درجة الحرارة‬‫الرطوبة‬‫وظيفة الذاكرة‬‫الحماية من الصدمة الكهربية‬‫درجة الحماية‬‫وضع التشغيل‬‫الحماية ضد دخول الماء‬‫األبعاد‬‫الوزن‬‫مدة الخدمة‪:‬‬‫الملحقات‬‫‪CH-650‬‬‫طريقة قياس الذبذبات‬‫شاشة رقمية‬‫‪ 13.5‬إلى ‪ 21.5‬سم‬‫‪ 0‬إلى ‪ 280‬مم زئبق‬‫‪ 40‬إلى ‪ 180‬نبضة‪/‬دقيقة‬‫‪ 3±‬مم زئبق‬‫‪ %5±‬من القراءة‬‫نفخ تلقائي بواسطة مضخة داخلية‬‫‪ :‬تيار مستمر)‬‫(‬‫‪ 3‬فولت تيار مستمر‬‫صمام تفريغ كهرومغناطيسي سريع‬‫بطاريتا ‪(LR03) AAA‬‬‫(‪ 180‬مم زئبق‪،‬‬‫حوالي ‪ 400‬مرة‬‫مرة‪/‬اليوم‪ °22 ،‬مئوية)‬‫‪ °10‬إلى ‪ °40‬مئوية‬‫‪ %15‬إلى ‪ %85‬من الرطوبة النسبية‬‫‪ °20‬إلى ‪ °60‬مئوية‬‫‪ %10‬إلى ‪ %95‬من الرطوبة النسبية‬‫حتى ‪ 60‬نتيجة قياس‬‫وحدة طاقة داخلية‬‫الجزء التطبيقي طراز ‪BF‬‬‫تشغيل مستمر‬‫‪IPX0‬‬‫‪( 65‬عرض) × ‪( 74‬ارتفاع) × ‪( 29‬عمق) مم‬‫حوالي ‪ 103‬جم بدون البطاريات‬‫‪ 5‬سنوات‬‫مجموعة تحتوي على بطاريتي ‪ (LR03) AAA‬لجهاز‬‫المراقبة‪ ،‬ودليل التعليمات‪ ،‬وصندوق التخزين‬‫العربية‬‫* الجزء العملي لهذا الجهاز هو الرباط‪.‬‬‫* نطاق بالضغط الجوي يتراوح من ‪ ٧٠٠‬هيكتوباسكال إلى ‪ ١٠٦٠‬هيكتوباسكال‬‫*تتوافق دقة النظام اإلجمالية لهذا الجهاز مع متطلبات البند ‪ 7.9‬من ‪.EN1060- 3‬‬‫‪ :EN1060- 3‬مقاييس ضغط الدم غير الباضعة الجزء ‪( 3‬المقاييس األوروبية)‬‫‪ -‬العربية ‪- 26‬‬‫‪2019/06/03 13:49‬‬‫‪CH-650̲1901̲Ara.indd 26‬‬‫قبل طلب اإلصالح أو االختبار‬‫تأكد من فحص ما يلي قبل إرسال جهاز مراقبة ضغط الدم لالختبار أو اإلصالح‪.‬‬‫المشكلة‬‫الفحوصات‬‫االستجابة‬‫استبدل كافة البطاريات بأخرى جديدة‪.‬‬‫تحقق مما إذا كانت البطاريات‬‫تظل الشاشة فارغة‬‫عندما تضغط على مفتاح فارغة‪.‬‬‫"بدء‪/‬إيقاف‬‫(‪ .")START/STOP‬تحقق مما إذا كانت البطاريات قم بتحميل البطاريات في االتجاه‬‫مركبة في االتجاهات الصحيحة الصحيح‪.‬‬‫( و موجهين بصورة‬‫صحيحة)‪.‬‬‫ال يمكن أخذ قياسات‪.‬‬‫تحقق مما إذا كانت أيقونة‬‫ظاهرة‪.‬‬‫القياس‬‫قم بتوصيل الرباط بصورة صحيحة‪.‬‬‫تحقق من توصيل الرباط‬‫بصورة صحيحة‪.‬‬‫هل بقيت ثاب ًتا أثناء القياس؟‬‫قم بقياس ضغط الدم لديك مرة أخرى‪،‬‬‫وتأكد من البقاء ثاب ًتا‪.‬‬‫الحظ أنه قد ال يمكن قياس ضغط الدم بالنسبة لشخص ما يعاني من‬‫ضعف حاد في النبض أو عدم انتظام ضربات القلب (نبض غير منتظم)‪.‬‬‫قراءات ضغط الدم لديك مرتفعة أو منخفضة بشكل غير‬‫ً‬‫مقارنة بتلك المأخوذة في المستشفى‪.‬‬‫طبيعي‬‫قم بإزالة البطاريات من الوحدة وتغييرها‬‫بأخرى جديدة‪.‬‬‫العربية‬‫ظاهرة أخرى‬‫انظر صفحة ‪.24‬‬‫‪ -‬العربية ‪- 25‬‬‫‪2019/06/03 13:49‬‬‫‪CH-650̲1901̲Ara.indd 25‬‬‫س و ج حول ضغط الدم‬‫س‪:‬‬‫ج‪:‬‬‫لماذا يختلف ضغط الدم المُقاس في المنزل عن ضغط الدم المُقاس بواسطة الطبيب في المستشفى؟‬‫		قد يختلف ضغط الدم لديك بسبب التمارين أو درجة الحرارة المحيطة أو الحالة الذهنية‪ .‬فعندما‬‫تقوم بقياس ضغط الدم لديك بواسطة طبيب أو ممرضة‪ ،‬ربما تكون قيمة الضغط أعلى بمقدار ‪10‬‬‫إلى ‪ 20‬مم زئبق عن الطبيعي نظرً ا للقلق و‪/‬أو اإلجهاد‪ .‬وتساعدك معرفة ما يؤثر على ضغط الدم‬‫في مراقبة صحتك بشكل أفضل‪.‬‬‫س‪:‬‬‫ج‪:‬‬‫لماذا يختلف ضغط الدم بأخذ القياسات؟‬‫		ضغط الدم لدينا مضبوط بوظيفة األعصاب التلقائية‪ .‬ويختلف ضغط الدم مع كل ضربة قلب‪.‬‬‫وربما نعتقد أن ضغط الدم لدينا ثابت‪ ،‬إال أنه يتغير إذا أخذت قياسات بصفة متكررة‪ .‬فضغط الدم‬‫يتأثر بتوقيت اليوم كالصباح والظهيرة‪ ،‬والفصول‪ ،‬ودرجة حرارة الجو‪ .‬وعالوة على ذلك‪ ،‬يتأثر‬‫ضغط الدم باإلجهاد الذهني أو الحاالت العاطفية المختلفة بسهولة‪ .‬فهو يميل لالرتفاع عندما تكون‬‫متوترً ا‪ ،‬ولالنخفاض عندما تكون مسترخيًا‪.‬‬‫س‪:‬‬‫		قياسات ضغط الدم المأخوذة في البيت تعطي بيانات موثوقة‪ ،‬إذ أنها تؤخذ عندما تكون في حالة‬‫ج‪:‬‬‫ما الفائدة من قياس ضغط الدم لدي في المنزل؟‬‫مستقرة ومسترخية‪ .‬ويولي األطباء اهتمامًا كبيرً ا لقياسات ضغط الدم المأخوذة في المنزل‪ .‬وينبغي‬‫عليك قياس ضغط الدم لديك في نفس الوقت من كل يوم‪ ،‬وتسجيل االختالفات في ضغط الدم لديك‪.‬‬‫استشر دائمًا طبيبك حول تفسير قراءات ضغط الدم لديك وتحديد العالج المناسب‪.‬‬‫العربية‬‫‪ -‬العربية ‪- 24‬‬‫‪2019/06/03 13:49‬‬‫‪CH-650̲1901̲Ara.indd 24‬‬‫حول "ضغط الدم"‬‫ما هو ضغط الدم؟‬‫القلب عبارة عن مضخة توزع الدم على الجسم بالكامل‪ .‬ويتم ضخ الدم من القلب بضغط مستمر إلى الشرايين‪ .‬هذا‬‫الضغط يُسمى ضغط الدم الشرياني ويمثل ضغط الدم لديك بصورة عامة‪ .‬ويُشار إلى ضغط الدم بعدة أنواع من‬‫الضغط‪ ،‬بما في ذلك الضغط االنقباضي الذي يحدث عندما يضخ القلب الدم‪ ،‬والضغط االنبساطي الذي يحدث عندما‬‫يعود الدم إلى القلب‪.‬‬‫الشريان‬‫(انبساطي)‬‫الشريان‬‫األُذين األيسر‬‫تتم استعادة الحالة‬‫العادية لجدار‬‫الشريان‪.‬‬‫انبساط‬‫(انقباضي)‬‫األُذين األيسر‬‫البُطين األيسر‬‫يتمدد جدار الشريان‪.‬‬‫البُطين‬‫األيسر‬‫انقباض‬‫يتغير ضغط الدم لديك طوال الوقت‪.‬‬‫يختلف ضغط الدم وف ًقا لس ّنك ونوعك وغير ذلك من العوامل‪ .‬وقد يتأثر بنظامك البيولوجي خالل اليوم وكذلك‬‫بوضعية جلوسك‪ ،‬واألنشطة الفيزيائية‪ ،‬واألنشطة العقلية‪ ،‬ومستوى اإلجهاد‪ ،‬وحتى بدرجة حرارة الجو‪ .‬ويتنوع‬‫ضغط الدم لدى الشخص الصحيح بصورة عامة خالل اليوم‪.‬‬‫[مثال لتنوعات ضغط الدم خالل اليوم]‬‫ضغط دم انقباضي‬‫‪160‬‬‫قيمة ضغط الدم (مم زئبق)‬‫‪140‬‬‫‪120‬‬‫‪80‬‬‫‪60‬‬‫ضغط دم انبساطي‬‫‪9 10 11‬‬‫‪40‬‬‫أثناء النوم‬
‫‪8‬‬

‫‪7‬‬

‫‪6‬‬

‫‪5‬‬

‫‪4‬‬

‫‪3‬‬

‫‪2‬‬

‫‪12 13 14 15 16 17 18 19 20 21 22 23 24 1‬‬‫التوقيت‬‫(قيم ضغط الدم مُقاسة بفواصل زمنية مقدارها خمس دقائق خالل اليوم)‬‫‪l Bevan AT, Honour AJ, Stott FH. Clin Sci 1969;36:329–44.‬‬‫‪ -‬العربية ‪- 23‬‬‫‪2019/06/03 13:49‬‬‫العربية‬‫‪100‬‬‫‪CH-650̲1901̲Ara.indd 23‬‬‫مؤشرات الخطأ‬‫عالمة الشاشة‬‫الحالة‪/‬السبب‬‫العربية‬‫تظهر عند لف رباط المعصم‬‫بصورة مرتخية للغاية‪.‬‬‫تظهر عند تحريك معصمك أثناء‬‫القياس‪.‬؛‬‫أو عند عدم لف الرباط بصورة‬‫صحيحة‪.‬‬‫أعِ د لف رباط المعصم بصورة‬‫صحيحة‪ .‬خذ قراءة أخرى‪ ،‬وتأكد‬‫من بقائك ثاب ًتا تمامًا‪.‬‬‫(‬‫انظر صفحة ‪" 14‬كيفية‬‫لف رباط المعصم"‪).‬‬‫تظهر عند نفاد البطاريات‪.‬‬‫استبدل البطاريتين القلويتين‬‫ببطاريتين جديدتين‪.‬‬‫(‬‫انظر صفحة ‪" 13‬إزالة‬‫البطاريات"‪).‬‬‫تظهر عندما يكون النبض خارج‬‫نطاق القياس (أدنى من ‪39‬‬‫ضربة‪/‬دقيقة أو أعلى من ‪181‬‬‫ضربة‪/‬دقيقة)‬‫أعِ د لف رباط المعصم بصورة‬‫صحيحة‪ .‬خذ أنفاسًا عميقة لكي‬‫تسترخي‪ ،‬ثم أعِ د القياس مرة أخرى‪.‬‬‫(‬‫انظر صفحة ‪" 16‬نصائح‬‫للحصول على قياسات دقيقة"‪).‬‬‫الوحدة ال تعمل بصورة سليمة‪.‬‬‫اتصل بالمتجر الذي اشتريت منه‬‫الوحدة أو بمركز الخدمة المحلي‬‫لديك‪( .‬‬‫انظر صفحة ‪25‬‬‫"قبل طلب اإلصالح أو االختبار"‪).‬‬‫‪ -‬العربية ‪- 22‬‬‫‪2019/06/03 13:49‬‬‫اإلجراء التصحيحي‬‫أعِ د لف رباط المعصم بصورة‬‫مريحة‪ .‬خذ قراءة أخرى‪ ،‬وتأكد من‬‫بقائك ثاب ًتا‪( .‬‬‫انظر صفحة‬‫‪" 14‬كيفية لف رباط المعصم"‪).‬‬‫‪CH-650̲1901̲Ara.indd 22‬‬‫استعادة البيانات المخزنة في الذاكرة‬‫‪2‬‬‫في كل مرة تقوم بالضغط على المفتاح‪ ،‬تظهر بيانات القياس بالترتيب من األحدث إلى األقدم‪.‬‬‫يظهر رقم البيانات قبل‬‫نتائج القياس‪.‬‬‫تظهر نتائج القياس بعد رقم البيانات‪.‬‬‫• القيمة المخزنة في الذاكرة مر ّقمة بترتيب القياسات‪ .‬على سبيل المثال‪ ،‬عند تخزين ‪ 60‬مجموعة‬‫من البيانات في الذاكرة‪ ،‬فإن البيانات رقم ‪ 1‬تمثل أحدث بيانات‪.‬‬‫‪60 ،59 ...‬‬‫رقم البيانات‪... ،3 ،2 ،1 :‬‬
‫‪Ç‬‬
‫‪Ç‬‬
‫أقدم بيانات‬‫أحدث بيانات‬‫اضغط على مفتاح "الذاكرة (‪.")M‬‬‫أثناء الضغط على مفتاح "الذاكرة (‪ ،")M‬اضغط على "بدء‪/‬إيقاف (‪")START/STOP‬‬‫ثوان‪.‬‬‫لمدة أكثر من ‪3‬‬‫ٍ‬‫سيتم حذف كافة البيانات‪.‬‬‫اضغط مع‬‫االستمرار لمدة‬‫ثوان‬‫‪3‬‬‫ٍ‬‫تم الحذف‬‫‪ -‬العربية ‪- 21‬‬‫‪2019/06/03 13:48‬‬‫تظهر رسالة تخبرك‬‫بأنه قد تم حذف كافة‬‫البيانات من الذاكرة‪.‬‬‫العربية‬

‫‪1‬‬
‫‪2‬‬

‫كيفية حذف البيانات‬‫‪CH-650̲1901̲Ara.indd 21‬‬‫قياس ضغط الدم لديك‬‫‪5‬‬‫إنهاء القياس‪.‬‬‫• اضغط على مفتاح "بدء‪/‬إيقاف (‪ ")START/STOP‬إليقاف تشغيل جهاز المراقبة‪.‬‬‫ميزة اإليقاف التلقائي‬‫• بعد أخذ قياسات ضغط الدم‪ ،‬سيؤدي الضغط على مفتاح "بدء‪/‬إيقاف (‪ ")START/STOP‬إلى‬‫إيقاف تشغيل جهاز المراقبة‪ .‬ومع ذلك‪ ،‬إذا لم تقم بالضغط على المفتاح‪ ،‬فستؤدي ميزة اإليقاف‬‫التلقائي بجهاز المراقبة إلى إيقاف تشغيله بعد ‪ 3‬دقائق‪.‬‬‫استعادة البيانات المخزنة في الذاكرة‬‫يمكن تخزين ‪ 60‬نتيجة قياس في الذاكرة‪.‬‬‫ُتحسب القيمة المتوسطة تلقائيًا لمساعدتك في إدارة صحتك اليومية‪.‬‬‫‪1‬‬‫كيفية استعادة البيانات‬‫اضغط على مفتاح "الذاكرة (‪.")M‬‬‫العربية‬‫• يظهر الرمز "‪ "3A‬الذي يشير إلى القيمة المتوسطة على الجزء األيسر من الشاشة‪ .‬ثم تظهر‬‫القيمة المتوسطة‪ ،‬بنا ًء على آخر ثالث قياسات‪( .‬ال يظهر "‪ "3A‬والقيمة المتوسطة في حالة‬‫وجود قراءتين أو أقل في الذاكرة)‪.‬‬‫‪ -‬العربية ‪- 20‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 20‬‬‫قياس ضغط الدم لديك‬‫‪4‬‬‫تظهر نتائج القياس‪.‬‬‫ضغط دم انقباضي‬‫ضغط دم انبساطي‬‫مؤشر عدم انتظام ضربات‬‫القلب (‪)IHB‬‬‫النبض‬‫مؤشر تحرك الجسم‬‫إذا تم اكتشاف ضربات قلب غير منتظمة أثناء القياس‪ ،‬فسيظهر مؤشر عدم‬‫انتظام ضربات القلب (‪ )IHB‬بعد اكتمال القياس‪.‬‬‫ُتعرّ ف ضربة القلب غير المنتظمة على أنها ضربة قلب تختلف بنسبة‬‫‪ %25±‬عن متوسط خمس فواصل ضربات قلب أثناء قياس ضغط الدم‪.‬‬‫• قد ال يكون القياس صحيحً ا إذا ترددت ضربات القلب بصورة كبيرة أثناء‬‫القياس‪ .‬إذا ظهر مؤشر عدم انتظام ضربات القلب (‪ ،)IHB‬فقم بقياس‬‫ضغط الدم لديك مرة أخرى‪.‬‬‫تحذير‬‫إذا ظهر مؤشر عدم انتظام ضربات القلب (‪ )IHB‬باستمرار‪،‬‬‫فنوصيك باستشارة طبيب حول حالتك الصحية‪.‬‬‫‪ -‬العربية ‪- 19‬‬‫‪2019/06/03 13:48‬‬‫العربية‬‫يظهر مؤشر تحرك الجسم على شاشة نتائج القياس إذا حرّ كت يدك أو‬‫معصمك أثناء القياس‪ ،‬مما يؤدي إلى اكتشاف تغير جوهري في الضغط‪.‬‬‫• إذا ظهر مؤشر تحرك الجسم‪ ،‬فقم بقياس ضغط الدم لديك مرة أخرى‪.‬‬‫‪CH-650̲1901̲Ara.indd 19‬‬‫قياس ضغط الدم لديك‬‫‪	1‬قم بلف رباط المعصم حول معصمك األيسر‪.‬‬

‫‪2‬‬
‫‪3‬‬

‫(‬

‫انظر صفحة ‪" 14‬كيفية لف رباط المعصم"‪).‬‬‫تأكد من أن وضعية جلوسك صحيحة‪.‬‬‫(‬‫انظر صفحة ‪" 16‬نصائح للحصول على قياسات دقيقة"‪).‬‬‫اضغط على مفتاح "بدء‪/‬إيقاف (‪ ")START/STOP‬لبدء‬‫القياس‪.‬‬‫• كافة األرقام الظاهرة عبارة عن عرض مبدئي‪ ،‬وهي‬‫تشير إلى أن جهاز المراقبة يعمل بشكل طبيعي‪.‬‬‫يتم أيضًا عرض العالمة‬‫ولكن ال يشير ذلك إلى‬‫ضرورة استبدال البطارية‪.‬‬‫يظهر‬‫عند اكتشاف نبض‪.‬‬‫العربية‬‫‪ -‬العربية ‪- 18‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 18‬‬‫نصائح للحصول على قياسات دقيقة‬‫للحصول على قياسات صحيحة‬‫• ال ُتطبق قبضتك بإحكام بالغ‪.‬‬‫مرتخ للغاية‪.‬‬‫• ال تقم بلف رباط المعصم بشكل‬‫ٍ‬‫• ال تغيّر موضع جهاز المراقبة‪.‬‬‫العربية‬‫• ال تحرّ ك جسمك وال تتحدث أثناء القياس‪.‬‬‫‪ -‬العربية ‪- 17‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 17‬‬‫نصائح للحصول على قياسات دقيقة‬‫يختلف ضغط الدم لديك وف ًقا لوضعية جلوسك وتوقيت القياس وغير ذلك من العوامل‪ .‬وبشكل‬‫مثالي‪ ،‬ينبغي عليك قياس ضغط الدم لديك بنفس وضعية الجلوس وفي نفس التوقيت من كل يوم‪.‬‬‫تأكد من أن وضعية الجلوس صحيحة‪.‬‬‫واسترخ‪.‬‬‫اجلس مستقي ًَما وخذ نفسًا عمي ًقا‬‫ِ‬‫وأبق‬‫اجعل راحة يدك موجهة ألعلى‪،‬‬‫ِ‬‫أصابعك مرتخية‪.‬‬‫من المهم أن يكون الرباط على‬‫مستوى واحد مع قلبك‪.‬‬‫العربية‬‫ضع منشفة مطوية أو شي ًئا مشابهًا تحت‬‫ذراعك‪.‬‬‫‪ -‬العربية ‪- 16‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 16‬‬‫كيفية لف رباط المعصم‬

‫‪4‬‬

‫‪5‬‬

‫ضع الحافة العلوية للرباط بعيدًا عن غضن المعصم بحوالي‬‫‪ 1‬إلى ‪ 2‬سم‪.‬‬‫أحكم ربط رباط المعصم بواسطة الشريط الالصق‪.‬‬‫• في حالة لف رباط المعصم بصورة مرتخية أو‬‫مشدودة للغاية‪ ،‬قد تنتج أخطاء أو قراءات غير‬‫صحيحة‪.‬‬‫للحصول على قياسات دقيقة‬‫• قد يؤدي لف الرباط بصورة مرتخية أو مشدودة للغاية إلى عدم الحصول على قياسات صحيحة‪.‬‬‫• اجلس في وضعية مستقيمة أثناء أخذ القياسات‪.‬‬‫‪ -‬العربية ‪- 15‬‬‫‪2019/06/03 13:48‬‬‫العربية‬‫يمكنك استخدام معصمك األيمن لقياس ضغط الدم لديك‪.‬‬‫ومع ذلك‪ ،‬قد يختلف ضغط الدم لديك بحوالي ‪ 10‬مم زئبق حسب معصمك األيمن أو األيسر‪ ،‬ولهذا‬‫ينبغي عليك‪ ،‬بشكل مثالي‪ ،‬قياس ضغط الدم لديك على نفس المعصم في كل مرة‪.‬‬‫‪CH-650̲1901̲Ara.indd 15‬‬‫كيفية لف رباط المعصم‬‫قم بلف رباط المعصم حول معصمك بصورة صحيحة‪ .‬قد يؤدي عدم لف الرباط بصورة سليمة إلى‬‫وجود أخطاء في القياس‪.‬‬

‫‪1‬‬
‫‪2‬‬

‫افتح رباط المعصم‪.‬‬‫قم بلف الرباط حول معصمك األيسر بحيث يكون‬‫جهاز المراقبة موضوعًا داخل معصمك‪.‬‬‫عار‪.‬‬‫• قم بلف رباط المعصم على جلد ٍ‬‫• رباط المعصم مُصمم بحيث يناسب المعصم الذي‬‫محيطه ‪ 13.5‬إلى ‪ 21.5‬سم‪.‬‬‫‪3‬‬‫اجعل الرباط مناسبًا لحجم معصمك بحريّة‬‫بسحب طرف الرباط‪.‬‬‫العربية‬‫‪ -‬العربية ‪- 14‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 14‬‬‫تحميل البطاريات‬‫‪3‬‬‫أغلق غطاء البطارية‪.‬‬‫قم بتحريك الغطاء في اتجاه السهم‪ .‬اضغط‬‫على الغطاء حتى يُصدر صوت طقطقة‪.‬‬‫إزالة البطاريات‬‫أو ال يظهر شيء على الشاشة‪ ،‬استبدل البطاريات القديمة بأخرى‬‫• عندما تظهر األيقونة‬‫جديدة‪ .‬يجب تغيير البطاريتين معًا في الحال‪.‬‬‫• استخدم البطاريات القلوية فقط‪.‬‬‫• ال تستخدم بطاريات قابلة إلعادة الشحن‪.‬‬‫• ال تستخدم بطاريتين إحداهما قلوية (‪ )LR03‬واألخرى بطارية منجنيز (‪ )R03‬معًا‪.‬‬‫• البطاريتان المضمنتان هما ألغراض توضيحية فقط‪ .‬قد يكون العمر االفتراضي للبطارية أقل‬‫من المحدد‪.‬‬‫العربية‬‫• عند التخلص من البطاريات‪ ،‬التزم باللوائح الحكومية أو قواعد المؤسسات العامة البيئية السارية‬‫في بلدك‪/‬منطقتك‪.‬‬‫‪ -‬العربية ‪- 13‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 13‬‬‫تحميل البطاريات‬‫(يجب تحميل البطاريات المرفقة مع الوحدة في حجيرتها قبل استخدام جهاز مراقبة ضغط الدم)‪.‬‬‫‪1‬‬‫افتح غطاء حجيرة البطاريات‪.‬‬‫غطاء البطارية‬‫مع الضغط على غطاء البطارية‬‫ألسفل‪ ،‬قم بتحريك الغطاء في اتجاه‬‫السهم‪.‬‬‫‪2‬‬‫ضع البطاريات في الحجيرة‪.‬‬‫ضع البطاريات مع األخذ في االعتبار رمزي القطبين الموجب والسالب‬‫* توضع البطاريات في االتجاه السالب أوالً‪.‬‬‫العربية‬‫أدخل البطارية القريبة من‬‫الشاشة أوالً‪.‬‬‫‪ -‬العربية ‪- 12‬‬‫‪2019/06/03 13:48‬‬

‫و‬

‫‪.‬‬

‫‪CH-650̲1901̲Ara.indd 12‬‬‫تعريف األجزاء‬‫<الشاشة>‬‫ضغط دم انقباضي‬‫ضغط دم انبساطي‬‫مؤشر عدم انتظام‬‫ضربات القلب (‪)IHB‬‬‫النبض‬‫العربية‬‫مؤشر مستوى شحن البطارية‬‫يظهر عندما تكون البطاريات منخفضة‪.‬‬‫(‬‫انظر صفحة ‪)13‬‬‫مؤشر تحرك الجسم‬‫‪ -‬العربية ‪- 11‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 11‬‬‫تعريف األجزاء‬‫أسماء المكونات‬‫<الوحدة الرئيسية>‬‫الشاشة‬‫مفتاح "الذاكرة (‪")M‬‬‫مفتاح "بدء‪/‬إيقاف‬‫)‪"(START/STOP‬‬‫انتظار‬‫غطاء البطارية‬‫رباط المعصم‬‫العربية‬‫‪ -‬العربية ‪- 10‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 10‬‬‫احتياطات االستخدام والصيانة‬CITIZEN SYSTEMS JAPAN CO.,LTD.
6-1-12, Tanashi-cho, Nishi-Tokyo-shi,
Tokyo 188-8511, Japan:‫االسم‬:‫العنوان‬‫المصنع‬CITIZEN SYSTEMS (JIANGMEN) CO., LTD.
:‫االسم‬Building 6, 399 Jinxing Road, Jianghai District, Jiangmen, Guangdong,	:‫العنوان‬China:‫المندوب باالتحاد األوروبي‬EMERGO EUROPE:‫االسم‬Prinsessegracht 20, 2514 AP The Hague, The Netherlands	:‫العنوان‬‫ يرجى االتصال بمندوبنا باالتحاد األوروبي‬،‫• للحصول على البيانات الفنية‬‫صُنع في الصين‬‫العربية‬CH-650̲1901̲Ara.indd 9- 9 ‫ العربية‬-2019/06/03 13:48

‫احتياطات االستخدام والصيانة‬‫طبيعة الصيانة وتكرارها‪:‬‬‫هذا المنتج مُصمم لالستخدام لفترة طويلة من الوقت؛ ورغم ذلك‪ ،‬يوصى عمومًا بفحصه‬‫كل عامين لضمان التشغيل واألداء السليمين‪.‬‬‫حماية البيئة الطبيعية‪:‬‬‫يُرجى المساعدة في حماية البيئة الطبيعية باتباع لوائح إعادة التدوير الوطنية و‪/‬أو المحلية عند التخلص من البطارية‬‫والمنتج في نهاية عمر استخدامه‪.‬‬‫عالمة ‪WEEE‬‬‫إذا أردت التخلص من هذا المنتج‪ ،‬فال تخلطه بالفضالت المنزلية العامة‪ .‬إذ توجد أنظمة تجميع‬‫منفصلة للمنتجات اإللكترونية المستعملة وف ًقا لقوانين بموجب توجيهات نفايات المعدات الكهربائية‬‫واإللكترونية (التوجيهات ‪ )2012/19/EU‬وهي سارية داخل االتحاد األوروبي فقط‪.‬‬‫معلومات حول التخلص في الدول األخرى خارج االتحاد األوروبي‪.‬‬‫إذا كنت ترغب في التخلص من البطاريات المستعملة‪ ،‬فيرجى االتصال بالسلطات المحلية لديك أو الموزع الخاص‬‫بك والسؤال عن الطريقة الصحيحة للتخلص منها‪.‬‬‫مالحظة حول رمز البطارية‪.‬‬‫قد يُستخدم الرمز باالشتراك مع رمز كيميائي‪ .‬وفي هذه الحالة‪ ،‬فهو متوافق مع المتطلبات المحددة بواسطة‬‫التوجيهات الخاصة بالمادة الكيميائية الموجودة‪.‬‬‫المعايير المرجعية األوروبية‪:‬‬‫يتطابق جهاز قياس ضغط الدم مع األنظمة ‪.EN1060-4 ،EN1060-3 ،EN1060-1 ،EN60601-1‬‬‫المعايرة‪:‬‬‫يوصى‪ ،‬بصفة عامة‪ ،‬بفحص جهاز المراقبة الخاص بقياس ضغط الدم كل عامين لضمان التشغيل الصحيح‬‫والدقة السليمة‪ .‬يرجى االتصال بالموزع‪.‬‬‫العربية‬‫‪ -‬العربية ‪- 8‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 8‬‬‫احتياطات االستخدام والصيانة‬‫تحذير‬‫شرح الرموز‬‫ال يسمح بتعديل هذه المعدات‪.‬‬‫‪:‬تهدف العالمة ‪ CE‬إلى توفير المعلومات لمفتشي السوق في البلدان األعضاء في‬‫االتحاد األوروبي‪.‬‬‫‪ :‬جهاز من النوع ‪BF‬‬‫‪ :‬أبقه جا ًفا‬‫‪ :‬تحذير‬‫‪ :‬ارجع إلى دليل التعليمات قبل االستخدام‬‫‪:‬توجيهات نفايات المعدات الكهربائية واإللكترونية المتعلقة بامتثال األجهزة‬‫‪ :‬جهة التصنيع‬‫العربية‬‫تحذير‬‫تم اختبار الجهاز واعتماده وف ًقا لمعيار ‪ EN60601-1-2‬الخاص‬‫بالتوافق الكهرومغناطيسي‪ .‬وال يضمن هذا‪ ،‬بأي شكل من األشكال‪،‬‬‫عدم تأثر الجهاز بالتداخل الكهرومغناطيسي‪ .‬ولذا‪ ،‬تجنب استخدام‬‫الجهاز في البيئات ذات الكهرومغناطيسية المرتفعة‪.‬‬‫‪ -‬العربية ‪- 7‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 7‬‬‫احتياطات االستخدام والصيانة‬‫‪ .13‬ال تستخدم األجهزة المحمولة بالقرب من الجهاز‪ ،‬فقد يؤدي هذا إلى تعطل الجهاز‪.‬‬‫‪ .14‬إذا كنت تستخدم مُنظمًا قلبيًا‪ ،‬فاستشر طبيبك قبل استخدام جهاز المراقبة‪.‬‬‫‪ .15‬تأكد من استخدام هذه الوحدة لقياس ضغط الدم فقط‪ .‬وال تستخدمها ألي غرض آخر‪.‬‬‫‪ .16‬تأكد من استخدام األجزاء والملحقات المرخصة فقط‪ .‬فقد تؤدي األجزاء والملحقات غير‬‫المرخص باستخدامها مع الجهاز إلى تلف الوحدة‪.‬‬‫‪ .17‬ال تستخدم هذه الوحدة على الرضع‪.‬‬‫‪ .18‬قد ال يمكن قياس ضغط الدم ألي شخص يعاني من حاالت شائعة لضربات القلب غير‬‫ُ‬‫المنتظمة مثل الضربات األُذينية أو البُطينية السابقة ألوانها أو الرجفان األذيني‪.‬‬‫احتياطات الصيانة‬‫‪.1‬‬‫‪.2‬‬‫‪.3‬‬‫‪.4‬‬‫‪.5‬‬‫‪.6‬‬‫‪.7‬‬‫‪.8‬‬‫ال تقم بتخزين جهاز مراقبة ضغط الدم في مواقع معرّ ضة لضوء الشمس المباشر‪ ،‬أو‬‫درجات الحرارة المرتفعة (أعلى من ‪ °60‬مئوية)‪ ،‬أو درجات الحرارة المنخفضة (أقل من‬‫‪ °20‬مئوية)‪ ،‬أو الرطوبة النسبية العالية (أعلى من ‪ ،)%95‬أو كميات الغبار الزائدة‪.‬‬‫تأكد من تخزين جهاز مراقبة ضغط الدم في مكان ال يتواجد فيه أطفال أو حيوانات أليفة أو‬‫حشرات ضارة‪.‬‬‫ال ُتسقط جهاز مراقبة ضغط الدم أو ُتعرّ ضه الهتزازات أو صدمات أخرى‪.‬‬‫قم بإزالة البطاريات إذا تركت الجهاز غير مُستخدم لفترة طويلة‪.‬‬‫ال تحاول فك الجهاز‪.‬‬‫ال تقم بثني الرباط بصورة زائدة عن الحد‪.‬‬‫ً‬‫متسخا للغاية‪ ،‬فامسحه بقطعة قماش مبللة بكحول معقم أو منظف متعادل‪ .‬ثم‬‫إذا كان الجهاز‬‫امسحه بقطعة قماش جافة‪.‬‬‫ال تقم أبدًا بتنظيف جهاز مراقبة ضغط الدم باستخدام تنر أو بنزين‪ ،‬فقد يؤدي هذا إلى‬‫تلف الجهاز‪.‬‬‫لتنظيف الرباط‪ ،‬امسحه بقطعة قماش مبللة‪ .‬تجنب الحك الخشن‪ ،‬فسوف يؤدي هذا إلى‬‫تسرب الهواء‪.‬‬‫العربية‬‫‪ -‬العربية ‪- 6‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 6‬‬‫احتياطات االستخدام والصيانة‬‫احتياطات االستخدام‬‫العربية‬‫‪ .1‬إذا كنت تعاني من مرض بالقلب أو ارتفاع ضغط الدم أو مرض آخر متعلق بالدورة الدموية‪،‬‬‫فاستشر طبيبك قبل استخدام جهاز المراقبة‪.‬‬‫‪ .2‬إذا شعرت بضغط غير عادي من الرباط أثناء االستخدام أو أي عرَ ض آخر غير طبيعي‪ ،‬فقلل‬‫الضغط من خالل فصل الطاقة أو إزالة البطاريات فورً ا‪ ،‬ثم ارجع إلى منفذ المبيعات الذي‬‫اشتريت منه الجهاز‪.‬‬‫‪ .3‬إذا رأيت أن القياس غير طبيعي أو يجعلك تشعر بأنك مريض‪ ،‬فتوقف عن استخدام الجهاز‬‫واستشر طبيبك‪.‬‬‫‪ .4‬قد ال يمكن قياس ضغط الدم ألي شخص يعاني من ضعف في النبض أو عدم انتظام ضربات القلب‪.‬‬‫‪ .5‬قد يؤدي القياس المتكرر لضغط الدم إلى مشكالت مثل االحتقان أو التورم عند بعض األشخاص‪.‬‬‫‪ .6‬لن تعطي القياسات المتكررة بصورة مستمرة لضغط الدم نتائج دقيقة‪ .‬ولذا‪ ،‬عليك أن تسمح‬‫بفاصل زمني مدته حوالي دقيقة بين كل عملية قياس‪.‬‬‫‪ .7‬إذا كنت تعاني من مشكلة حادة في الدورة الدموية في ذراعيك‪ ،‬فاستشر طبيبك قبل استخدام‬‫الجهاز‪ .‬فقد يكون عدم القيام بذلك خطرً ا على صحتك‪.‬‬‫‪ .8‬قد ال يمكن القياس ألي شخص يعاني من ضعف ضخ الدم إلى المنطقة التي سيتم فيها القياس‬‫أو شخص يعاني من ضربات قلب غير منتظمة بصورة دائمة‪ .‬استشر طبيبك واطلب نصيحته‬‫حول استخدام الجهاز من عدمه‪.‬‬‫‪ .9‬ال تقم بلف الرباط حول معصم مجروح‪.‬‬‫‪ .10‬ال تقم بلف الرباط حول معصم به محقن أوردة‪ ،‬أو معصم مستخدَ م في نقل الدم كجزء من‬‫عالج طبي‪ .‬فيمكن أن يؤدي القيام بذلك إلى اإلصابة أو إلى حادث خطير‪.‬‬‫‪ .11‬ال تستخدم الجهاز بالقرب من الغازات القابلة لالشتعال مثل الغازات المُستخدمة في التخدير‪.‬‬‫فيمكن أن يؤدي القيام بذلك إلى إشعال الغازات والتسبب في وقوع انفجار‪.‬‬‫‪ .12‬ال تستخدم الجهاز في بيئات غنية باألوكسجين مثل غرفة الضغط العالي أو خيمة األوكسجين‬‫بالمستشفيات‪ .‬فيمكن أن يؤدي القيام بذلك إلى إشعال األوكسجين والتسبب في نشوب حريق‪.‬‬‫‪ -‬العربية ‪- 5‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 5‬‬‫مالحظات عامة‬‫‪ .10‬إذا ظهر مؤشر تحرك الجسم أو مؤشر عدم انتظام ضربات القلب (‪ )IHB‬على الشاشة‪،‬‬‫فاسترخ وقم بقياس ضغط الدم مرة أخرى‪ .‬إذا ظهر مؤشر عدم انتظام ضربات القلب (‪)IHB‬‬‫ِ‬‫باستمرار‪ ،‬فنوصيك باستشارة طبيب حول حالتك الصحية‪( .‬‬‫انظر صفحة ‪)19‬‬‫ً‬‫‪ .11‬الشكل التالي يوضح تصنيفات ضغط الدم وفقا لمعايير منظمة الصحة العالمية‪.‬‬‫انقباضي‬‫(مم زئبق)‬‫‪180‬‬‫الدرجة ‪ 3‬ارتفاع ضغط الدم (حاد)‬‫الدرجة ‪ 2‬ارتفاع ضغط الدم (متوسط)‬‫‪160‬‬‫‪140‬‬‫‪130‬‬‫‪120‬‬‫الدرجة ‪ 1‬ارتفاع ضغط الدم‬‫(طفيف)‬‫مالحظات‪:‬‬‫الرسم ليس دقي ًقا‪ ،‬ولكن يمكن استخدامه‬‫كمرشد في فهم قياسات ضغط الدم غير‬‫الباضعة‪ .‬الجهاز مُصمم لالستخدام من‬‫قبل البالغين فقط‪.‬‬‫عال‪-‬عادي‬‫ٍ‬‫عادي‬‫مثالي‬‫‪90 85 80‬‬‫‪100‬‬‫‪110‬‬‫انبساطي‬‫(مم زئبق)‬‫تعريفات وتصنيفات مستويات ضغط الدم‬‫‪ .12‬الغرض من االستخدام‬‫• هذا الجهاز هو غير باضع ومخصص لمراقبة ضغط الدم بواسطة طريقة قياس الذبذبات‪.‬‬‫يمكن لهذا الجهاز قياس ضغط الدم االنقباضي (‪ ،)SYS‬وضغط الدم االنبساطي (‪،)DIA‬‬‫ومعدل النبض تلقائيا‪.‬‬‫• تم تصميم هذا الجهاز الستخدامه لفحص الحالة الصحية الشخصية في المنزل تحت إشراف‬‫ً‬‫جهازا تشخيصيًا‪.‬‬‫طبيب ولم يتم تصميمه ليكون‬‫العربية‬‫• لم يتم تصميم هذا الجهاز لالستخدام مع األطفال حديثي الوالدة أو الرضع‪ ،‬ولم يتم تصميمه‬‫لقياس الدوران التلقائي‪.‬‬‫قم بقياس ضغط الدم لديك في نفس التوقيت من كل يوم‪.‬‬‫* يتغير ضغط الدم لديك طوال الوقت‪ .‬ويعني هذا أن البيانات التي يتم تجميعها على مدى فترة‬‫طويلة لها أهمية أكبر من البيانات الناتجة من قياس واحد‪ .‬ولهذا‪ ،‬يجب عليك قياس ضغط الدم‬‫لديك بشكل يومي‪ .‬وبشكل مثالي‪ ،‬ينبغي عليك قياس ضغط الدم لديك في نفس التوقيت من كل‬‫يوم‪ ،‬حيثما أمكن‪.‬‬‫‪ -‬العربية ‪- 4‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 4‬‬‫مالحظات عامة‬‫مالحظات عامة حول ضغط الدم وقياسه‬‫‪ .1‬لتفريغ هواء الرباط بسرعة‪ ،‬اضغط على مفتاح "بدء‪/‬إيقاف (‪.")START/STOP‬‬‫‪ .2‬الرباط المرفق مناسب للمعصم الذي محيطه ‪ 21.5-13.5‬سم‪.‬‬‫‪ .3‬عند إرفاق الرباط‪ ،‬يجب وضع جسم جهاز مراقبة ضغط الدم داخل المعصم‪.‬‬‫ً‬‫جهازا تشخيصيًا‪ ،‬بل هو مُصمم لالستخدام المنزلي فقط‪.‬‬‫‪ .4‬الجهاز غير مُصمم بحيث يكون‬‫‪ .5‬القياس الذاتي ليس عالجً ا! ولذا ينبغي عليك عدم تغيير الدواء الموصوف لك من الطبيب في‬‫أي حال من األحوال‪.‬‬‫‪ .6‬كتحضير لقياس ضغط الدم لديك‪ ،‬يجب عليك أن تتبول ثم تظل ثاب ًتا نسبيًا لمدة ‪ 10‬إلى ‪15‬‬‫دقيقة قبل القياس‪.‬‬‫‪ .7‬فالتمرين واألكل والشرب والتدخين وما إلى ذلك قبل القياس قد يؤثر على النتائج‪.‬‬‫‪ .8‬يختلف ضغط الدم لديك باستمرار طوال اليوم‪ .‬ولذا‪ ،‬يجب أخذ القياسات بصورة منتظمة كل‬‫يوم في نفس التوقيت‪.‬‬‫‪ .9‬ال تضع تركيزك كله على نتائج قياس واحد‪ .‬بل استمر في تسجيل النتائج المتنوعة لضغط الدم‬‫لديك‪ .‬إذ يمكنك الحصول على صورة كاملة من خالل عدد كبير من القراءات‪.‬‬‫‪ .10‬يؤدي اإلجهاد العاطفي إلى رفع ضغط الدم‪.‬‬‫‪.1‬‬‫‪.2‬‬‫‪.3‬‬‫‪.4‬‬‫‪.5‬‬‫‪.6‬‬‫‪.7‬‬‫‪.8‬‬‫‪.9‬‬‫‪ -‬العربية ‪- 3‬‬‫‪2019/06/03 13:48‬‬‫العربية‬‫للحصول على قياسات ضغط دم صحيحة‬‫استرخ قبل قياس ضغط الدم لديك‪ .‬إذا كنت متوترً ا عند أخذ‬‫خذ خمسة أو ستة أنفاس عميقة ثم‬‫ِ‬‫القياس‪ ،‬فلن تحصل على قراءة صحيحة‪.‬‬‫سيرتفع ضغط الدم لديك إذا كنت قل ًقا أو منزعجً ا‪ ،‬أو كنت تعاني من قلة النوم أو اإلمساك‪ ،‬أو‬‫إذا كنت قد أجريت بعض التمارين أو تناولت وجبة‪.‬‬‫انظر صفحة ‪)14‬‬‫قم بلف الرباط حول معصمك بصورة مريحة‪( .‬‬‫ال تقم بقياس ضغط الدم لديك بعد االستحمام أو الشرب‪.‬‬‫إذا شعرت أنك بحاجة للتبول‪ ،‬فقم بذلك قبل قياس ضغط الدم لديك‪.‬‬‫قم بقياس ضغط الدم لديك عندما تكون درجة حرارة الغرفة حوالي ‪ °20‬مئوية‪ .‬وال تقم بقياس‬‫ضغط الدم لديك عندما تكون درجة حرارة الغرفة باردة للغاية (أدنى من ‪ °10‬مئوية) أو‬‫ساخنة للغاية (أعلى من ‪ °40‬مئوية)‪.‬‬‫ال تحاول قياس ضغط الدم لديك بعد شرب القهوة أو الشاي أو بعد التدخين مباشر ًة‪.‬‬‫أبق مركز الرباط في مستوى القلب‬‫قم بقياس ضغط الدم لديك عندما تكون مسترخيًا وثاب ًتا‪ِ .‬‬‫وال تحرّ ك ذراعك وال تتحدث‪.‬‬‫ال تقم بقياس ضغط الدم لديك عندما يكون الرباط ملفو ًفا على معصمك منذ بضع دقائق أو أكثر‪.‬‬‫فخالل ذلك الوقت‪ ،‬سيعمل ذراعك على رفع ضغط الدم ولن تحصل على قراءة صحيحة‪.‬‬‫‪CH-650̲1901̲Ara.indd 3‬‬‫المحتويات‬‫‪3‬‬‫‪ -‬مالحظات عامة‬‫‪5‬‬‫‪ -‬احتياطات االستخدام والصيانة‬‫‪ - 10‬تعريف األجزاء‬‫‪ - 12‬تحميل البطاريات‬‫‪ - 13‬إزالة البطاريات‬‫‪ - 14‬كيفية لف رباط المعصم‬‫‪ - 16‬نصائح للحصول على قياسات دقيقة‬‫‪ - 18‬قياس ضغط الدم لديك‬‫‪ - 20‬استعادة البيانات المخزنة في الذاكرة‬‫‪ - 22‬مؤشرات الخطأ‬‫‪ - 23‬حول "ضغط الدم"‬‫‪ - 24‬س و ج حول ضغط الدم‬‫‪ - 25‬قبل طلب اإلصالح أو االختبار‬‫‪ - 26‬المواصفات‬‫تأكد من أن لديك كافة المكونات التالية‬‫العربية‬‫وحدة جهاز مراقبة‬‫ضغط الدم‬‫صندوق التخزين‬‫(خاصة بجهاز المراقبة)‬‫بطاريتا ‪AAA‬‬‫‪ -‬العربية ‪- 2‬‬‫‪2019/06/03 13:48‬‬‫دليل التعليمات‬‫‪CH-650̲1901̲Ara.indd 2‬‬‫دليل تعليمات‬‫جهاز‬‫مراقبة ضغط الدم الرقمي‬‫العربية‬‫‪CH-650‬‬‫ﺍﻟﻌﺮﺑﻴﺔ‬‫‪1901‬‬‫‪2019/06/03 13:48‬‬‫‪CH-650̲1901̲Ara.indd 1‬‬