TELS

41

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS
introduction
The Mini-Horizontal TELS is the ideal instrument for checking
small dimensions. It is part of the horizontal instruments family.
The instrument is mainly used for checking internal and external dimensions, such as ring gauges, plug gauges, threadplug gauges and for measuring precision production parts.
The reliable concept is easily suitable for the use in workshop environment.
An electronic probe and digital display unit or simply a dial indicator can be applied to perform all measurements. The selectionof the measuring system depends on the required accuracy.
The range for internal measurements from 10 to 100 mm andexternal measurements from 0 to 100 mm covers a large number of applications. The absolute measuring range is 25 mm
and a preset setting is necessary to obtain an entire length.
A wide range of accessories are available.
IDEAL FOR CHECKING SMALL DIMENSIONSEASY MANIPULATIONSUITABLE FOR USE IN WORKSHOP AREA
WIDE SELECTION OF ACCESSORIES

42

DESCRIPTIONInterchangeable inserts forexternal measurements

Adjustable datum
supportSupport table,
adjustable in height

Knurled knob to movethe probe guiding device

Interchangeable anvils forinternal measurements

Axis for adjustment of
measuring force

43

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS
display / software

FUNCTIONS OF SYLVAC DISPLAY UNIT:
SELECTION OF THE MEASURING UNIT MM/INCH
ANALOG DISPLAYINPUT AND DISPLAY OF TOLERANCE LIMITS
CLASSIFICATION (2 - 6 CLASSES)INPUT AND SETTING OF PRESET VALUESINVERSION OF MEASURING DIRECTION (+/-)
CHANNEL SELECTIONVALUE HOLD
LOCKING OF THE KEYBOARD AND THE MM/INCHSELECTION
EXTERNAL CONTACT FUNCTION USING FOOT PEDALOR COMPUTERPRINT-OUT OF VALUE AND STATISTICAL ANALYSIS

technical specifications
TELS
Absolute measuring range

mm (in)25 (1)

Application range (external measurements)

mm (in)100 (4)

Application range (internal measurements)

mm (in)100 (4)Max. permissible errors (according to measuring probe) 1)

µm

1.5

Repeatability (2s)

µm

0.1

mm (in)0.1 ÷ 0.0001 (.001÷ 00001)

Measuring force (adjustable)N

3-8

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

WeightKg

15

Resolutions

1)

1)

1)

Depends on measuring probe. Values valid with probe P25 at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5 %.

44

105.7

112.7

100

136

42

39.8

SCHEMA

365

99

390

163

standard instrument
The TELS instruments are supplied as follows:
Instrument according to specifications

User’s manual (750 50 0019 03)

Measuring inserts for external measurement (TELS50)

Lapping plate (TA-TO-301)

Measuring anvils for internal measurements (TELS10)

Test certificate

CODE NUMBERs
TELS
TELS 700 205 00 01

45

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS
Applications

Checking of external diameters(TELS50, TELS5, TELS5.1)

Checking of lengths(TELS50, TELS5, TELS5.1)

46

Checking of internal diameters(TELS10)Checking of internal grooves(TELS10)

ALESTA47

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

ALESTA
introduction
The Alesta family is a product that simplifies the setting of
2-point bore gauges. It replaces the traditional way using a
lot of different rings.
Thanks to innovative adapters, a positioning error will beeliminated and the speed of measurements increased. You
just input the dimension into the key board and the required
mobile carriage will position itself automatically onto this position. The mobile carriage system will always stay in the desired position even when applying a force on the anvils.
The stable granite base combined with a high precision measuring system guarantees excellent measuring results. Using a tolerance table the carriage can be positioned to theselected tolerance range. This table can be tailor-made tocustomer’s requirements. This product achieves a typical return on investment of 2 years. The acquisition of new rings
and their calibration will be eliminated. The instrument range
includes 3 models: 300, 500 and 1000 mm.
IDEAL FOR USE IN THE WORKSHOP AREA
VERY SIMPLE MANIPULATIONMOTORIZED CARRIAGE MOVEMENTSTABLE GRANITE BASE
HIGH PRECISION MEASURING SYSTEM (HEIDENHAIN)TOLERANCE DATA BASE
WIDE SELECTION OF ADAPTERS AND ACCESSORIESINTERNAL AND EXTERNAL MEASUREMENT POSSIBLE

48

DESCRIPTION

Display unit

Granite base

Mobilemeasuring anvil

Fixed measuring anvil

Support to fix the adaptors

49

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

ALESTA
display / software

PROGRAMMING OF 100 TOLERANCED DIMENSIONSMANUAL OR AUTOMATIC DISPLACEMENTPROGRAM IN ITALIAN, ENGLISH, FRENCH,PORTUGUESE, GERMAN, SPANISH
INTERNAL AND EXTERNAL MEASUREMENT

technical specifications
AlestaApplication range (external measurements) mm (in)Application range (internal measurements)

mm (in)

300

500

1000

300 (12)500 (20)1000 (40)40 ÷ 340 (1.6 ÷ 13.5)

40 ÷ 540 (1.6 ÷ 21.5)

40 ÷ 1040 (1.6 ÷ 41)µm

1.5 + L(mm) / 300µm<1Resolutions

mm (in)0.001 (.00001)

Holding force

N

Max. permissible errors

1)

Repeatability (2s) 1)

240

Interfaces

1)

RS232 et USB

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

WeightKg

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

50

62

77

200

247 / 247 / 340

SCHEMA

300 / 300 / 330

610 / 820 / 1330

standard instrument
The Alesta instruments are supplied as follows:
Instrument according to specifications

User’s manual (750 50 0022 00)
Test certificate

CODE NUMBERs
Alesta
ALE300P 700 204 00 02

Measuring range 300 mm

ALE500P 700 204 10 02

Measuring range 500 mm

ALE1000P 700 204 20 02

Measuring range 1000 mm

51

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

ALESTA
Applications

Setting of two-point bore gauges

No need to have a lot of different setting rings. They are replaced by one
single instrument

Wide range of adapters to fit most
two-point bore gauges and garanteeof easy / quick setting

Quick positioning using
pre-programmed data

52

Checking of calipers

HORIZON+ Granite53

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON + Granite
introduction
The «Horizon» instruments respond to the current requirements for quality inspection in manufacturing areas. They allow a quick, precise checking and setting.
The instruments are perfect for setting all kinds of comparative measuring equipment and for checking length, internaland external diameters, thread gauges etc.
The new innovative concept with incorporated measuring system and display unit ensures excellent results by easy manipulation and is the answer to various demands in the field of production. Simplicity of use is an additional valuable advantage.
The instrument range comprises several models, from 500
mm up to 2000 mm (Granite 3000 - 8000 mm). A wide selection of easy interchangeable accessories completes the application possibilities.
IDEAL FOR USE IN THE WORKSHOP AREA
VERY SIMPLE MANIPULATIONPRECISE CHECKING AND SETTING MADE EASY
SELF-CONTAINED OPERATION OF APPROX. 100 HOURS
WIDE SELECTION OF ACCESSORIES54

DESCRIPTION
HorizonInterchangeable
accessories

Dimensionally stable instrumentbase, positioned on threelevel points

Digital display with
RS232 data output

Locking device (locking of
the measuring carriage)

Handwheel for the
displacement of themeasuring carriage

Horizon Granite

55

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON + Granite
display / software
Horizon

SELECTION OF VALUE: NORMAL / MIN / MAXZERO SETTING OF THE DISPLAY AND PRESET INPUT
RS232 DATA OUTPUTTWO REFERENCES
SELECTION OF THE RESOLUTION 0.01 MM - 0.001 MM
SELECTION OF MEASURING UNIT MM/INCH (DIRECTCONVERSION)

Horizon Granite

Heidenhain ND287 or ND1100
LINEAR MEASUREMENT DISPLAY AND MIN. / MAX.
VALUE HOLD
DIGITAL DISPLAY
ZERO SETTING OF THE DISPLAY AND PRESET INPUTPARAMETER SETTINGSINITIALIZATION OF THE DISPLAY USING EXTERNALFUNCTION
RS232 DATA OUTPUT

PC with software WinDHI
GRAPHIC HELP FOR MEASURING FUNCTIONSSELECTION OF THE MEASURING UNIT MM / INCH
DIRECT DISPLAY OF ALL LENGTH MEASURING VALUESAND MIN. / MAX. VALUE HOLD
INPUT OF 9 PRESET VALUESINVERSION OF MEASURING DIRECTION SIGN (+/-)
DATA TRANSFER USING A FOOT PEDAL

56

Heidenhain ND287

Heidenhain ND1100

TECHNICAL SPECIFICATIONS
HorizonMeasuring range

mm (in)

H501

H1001

520 (20)1020 (40)1520 (60)2020 (80)µm

2.5 + L/300

Repeatability (2s) 1)µm

1Resolutions

mm (in)0.01/0.001 (.0001 / .00005)

Max. displacement speed

mm  / s1500

Measuring force

N

3

Autonomyh100

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weight

Kg

RS232

94

126

158

190

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

Horizon Granite
Measuring range

mm (in)

HG3000

HG4000

HG5000

HG6000

HG80003000 (118)4000 (157)5000 (196)6000 (236)8000 (315)µm

0.7 + L/750

µm

0.5

Resolutions

mm (in)0.01 / 0.001 / 0.0001 (.001 / .0001 / .00001)

Max. displacement speed

mm  / s

2000

N

3

Max. permissible errors

1)

Repeatability (2s) 1)

Measuring force
Interfaces

1)

H2001

Max. permissible errors 1)

Interface

1)

H1501

RS232/USB

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Relative humidity%

20 ÷ 80

Weight

Kg

1060

1980

3200

3700

4600

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

57

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON + Granite
SCHEMA295~ 417

Horizon554 / 831 / 1108 / 1385

270

1012 / 1512 / 2012 / 2512

970

450/450/550/
650/850

Horizon Granite

2660/2660/3220/3780/4900

390

4750/4750/5750/6750/8750

58

STANDARD INSTRUMENTThe Horizon instruments are supplied as follows:
Instrument according to specifications
Pair of parallel attachments 25 mm (TEL5)
Universal AC adapter (TA-EL-131)
Protection cover (TEL.HO500 / 1000 / 1500 / 2000)
Allen key set (TA-TO-004)
User’s manual (750 50 0005 03)
Test certificate

The Horizon Granite instruments are supplied as follows:
Instrument according to specifications
Pair of anvils with tungsten carbide surface (TEL1)
Lapping plate (TA-TO-302)
Protection cover (TEL.HO3000 / 4000 / 5000 / 6000 / 8000)
Compressed air tubing and air filter
Allen key set (TA-TO-004)
User’s manual (750 50 0003 03)
Test certificate

CODE NUMBERs
Horizon
H501 700 201 10 02

Measuring range 500 mm

H1001 700 201 20 02

Measuring range 1000 mm

H1501 700 201 30 02

Measuring range 1500 mm

H2001 700 201 40 02

Measuring range 2000 mm

Horizon Granite
HG3000 700 208 60 11

Measuring range 3000 mm

HG4000 700 208 70 11

Measuring range 4000 mm

HG5000 700 208 80 11

Measuring range 5000 mm

HG6000 700 208 90 11

Measuring range 6000 mm

HG8000 700 208 110 11

Measuring range 8000 mm

59

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON + Granite
Applications

Setting of 2-point internal and external
comparative measuring equipment(TEL5 / TELMA7 / TELMN7.2)

Setting of 2-point bore gauges(TEL5 / TA-SU-301)

Checking of calipers (TEL5.10)

Checking of snap gauges (TELMA8)

Checking of internal diameter ofheavy rings (TEL9 / H-5)

Checking of depth gauges(TULM19 / TEL5)

Checking of external micrometers(TEL5 / TEL11)

Checking of dial indicators (TEL5CN)

Checking of 2-point internal micrometers (TEL5 / TELMA7 / TELMN7.2)

60

HORIZONPREMIUM

61

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON PREMIUM
INTRODUCTION
The HORIZON PREMIUM instruments fulfill today’s requirements of high accuracy quality control equipment in production
areas and take into consideration EN ISO 9000 standards.
The instruments may be used in a workshop area or installed
in a quality control room for checking of measuring gauges
and setting of all types of comparative measuring equipment.
The well-designed, reliable new concept, obtained by superior
engineering, ensures high precision and optimum results byeasy manipulation as well as a valuable level of productivity.
The modular design allows the selection between two measuring systems: analog signal output and Heidenhain display
unit or digital signal output and computer with TFT touch
screen and Trimos-WinDHI software.
Instruments with a measuring range from 500 to 3000 mm areavailable, all of them built in one single piece. All measuring
ranges being direct, it means that the whole measuring range
is available without adjustment or intermediate re-calibration.
IDEAL FOR THE USE IN LABORATORY AND IN
THE WORKSHOPNEW ERGONOMIC CONCEPTMEASURING CARRIAGE WITH FINE ADJUSTMENTLARGE RANGE OF ACCESSORIESADJUSTABLE MEASURING FORCE
3 DISPLAYS AVAILABLE
MODULAR SYSTEM DESIGNDIRECT MEASUREMENT OVER THE WHOLE RANGE

62

DESCRIPTION

HPA
Display Heidenhain ND1100
or ND287Interchangeable
accessories

Digital display of the set
measuring force in Newton

Dimensionally stable instrument basepositioned on three level points(two additional balance points).

HPD
Touch screen with
TRIMOS-WinDHI software
for all required measuringfunctions

Measuring carriage withhandwheel for displacement,
locking and fine adjustement

63

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON PREMIUM
display / software
Horizon Premium (HPA)

Heidenhain ND287 or ND1100 display units
LINEAR MEASURING SYSTEM, MIN. / MAX. VALUE HOLD
DIGITAL DISPLAY
ZERO SETTING OF THE DISPLAY AND PRESET INPUTPARAMETER SETTINGS AND CLASSIFICATIONCONFIGURATION OF THE DISPLAY USING EXTERNALCONTACT
RS232 DATA OUTPUT

Horizon Premium (HPD)

TRIMOS WinDHI
TRIMOS-WinDHI Software allows the performance of all required measuring functions and the connection to the temperature compensation system TempComp as well as to agauge inspection and management system.
PC WITH TOUCH SCREEN
DDE-SERVER (FOR EXCEL, WORD, ETC.)
GRAPHIC HELP FOR MEASURING FUNCTIONSDATA TRANSFER USING A FOOT PEDALDIGITAL DISPLAY OF THE SELECTED MEASURING
FORCE IN NEWTON (N)DIRECT DISPLAY OF ALL LENGTH MEASURING VALUESAND MIN. / MAX. VALUE HOLD
MEASURING WITH 9 REFERENCESINVERSION OF MEASURING DIRECTION SIGN (+/-)
COMPATIBLE WITH TEMPERATURE COMPENSATIONSYSTEM TEMPCOMP (SEE § LABCONCEPT)QMSOFT

Trimos recommends the QMSOFT software package for inspection and management of all measuring tools available.
INTEGRATED DRIVERS FOR TRIMOS INSTRUMENTSREQUIRED NOMINAL SIZES AND TOLERANCESAVAILABLE ACCORDING ALL STANDARDSCUSTOMIZED INSPECTION CERTIFICATES

64

Heidenhain ND1100

Heidenhain ND287

TECHNICAL SPECIFICATIONS
Horizon Premium
Measuring range

mm (in)

1000

1500

2000

3000

550 (21)1050 (41)1550 (61)2050 (80)3050 (120)µm

0.7 + L (mm)/1000

µm

0.2

Resolutions

mm (in)0.01 / 0.001 / 0.0001 (.001 / .0001 / .00001)

Max. displacement speed

mm/s

1.5

Max. permissible errors

1)

Repeatability (2s) 1)

1)

500

Measuring forceN

0 ÷ 12

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Relative humidity%

20 ÷ 80

Weight

kg

94

123

152

181

239

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

427

SCHEMA616 / 892 / 1170 / 1446 / 2000

270

1122 / 1622 / 2122 /2622 / 3622

65

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON PREMIUM
STANDARD INSTRUMENTThe Horizon Premium (HPA) instruments are supplied as follows:
Instrument according to specifications
Pair of anvils with tungsten carbide surface (HPA-1)
Lapping plate (TA-TO-302)
Protection cover (TEL.HO500 / 1000 / 1500 / 2000)
Allen key set (TA-TO-004)
User’s manual (750 50 0015 03)
Test certificate

The Horizon Premium (HPD) instruments are supplied as follows:
Instrument according to specifications
Pair of anvils with tungsten carbide surface (HPA-1)
PC with interface, touch screen 1), adjustable support 1), touch screen pen 1)
Foot pedal for data transfer (TELMA31)
Opto-RS connection cable for measuring force (TVM.O-PC/AT.9P)
Lapping plate (TA-TO-302)
Protection cover (TEL.HO500 / 1000 / 1500 / 2000)
Allen key set (TA-TO-004)
User’s manual (750 50 0015 03)
Test certificate
1)

Not included in version HPD-B. Touch screen replaced by a regular TFT screen.

CODE NUMBERs
Horizon Premium (HPA)
HPA500 700 202 10 01

Measuring range 500 mm

HPA1000 700 202 20 01

Measuring range 1000 mm

HPA1500 700 202 30 01

Measuring range 1500 mm

HPA2000 700 202 40 01

Measuring range 2000 mm

HPA3000 700 202 50 01

Measuring range 3000 mm

Horizon Premium (HPD)
HPD500 700 212 10 01

HPD500B 700 212 10 02 1)

Measuring range 500 mm

HPD1000 700 212 20 01

HPD1000B 700 212 20 02

1)

Measuring range 1000 mm

HPD1500 700 212 30 01

HPD1500B 700 212 30 02 1)

Measuring range 1500 mm

HPD2000 700 212 40 01

HPD2000B 700 212 40 02 1)

Measuring range 2000 mm

HPD3000 700 212 60 01

HPD3000B 700 212 60 02

Measuring range 3000 mm

1)

HPD-B: With TFT screen, no adjustable support.

66

1)

APPLICATIONSChecking of ring gauges(TA-SU-313 / TEL16.1 / HPA-1)

Checking of plug gauges(HPA-1 / TULM6 / L05 / LABC-15)

Checking of calipers (TEL5.10)

HPD: Checking of thread ring gauges(TA-SU-313 / LABC-70 / TA-SU-354)

Checking of thread plug gauges (HPA1 / TEL6 / 3P / 0.17-3.2 / S6.5 / LABC-15)

Checking of external micrometers(HPA-1 / TULM14)

Checking of snap gauges(TA-SU-313 / TEL14N)

Checking of dial indicators (TEL5CN)

HPD: Checking of taper thread ringgauges (LABC80)

67

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

HORIZON PREMIUMAPPLICATIONS

Setting of 2-point internal micrometers(HPA-1 / TELMA7 / TELMN7.2)

Checking of depth gauges(TEL5 / TULM19)

HPA: Measurement of cylindricalparts (TA-MS-303)

Large micrometer control(TEL5 / TEL11 / LABC20)

Setting of a large gauge (TEL5)

68

HPA: Checking of thread ring gauges(TEL18)

Setting of 2-point bore gauges(TEL5 / TA-SU-301)

THV

69

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

THV
introduction
The horizontal THV instrument has been designed for calibration and certification of gauging equipment of small dimension.
It allows easy and precise checking of plug gauges, ringgauges, thread plug gauges, test and dial indicators as wellas measuring of high precision production parts.
The incorporated opto-electronic measuring system guarantees high accuracy. A separate display unit or a PC with
Trimos WinDHI software can be used to display the measurement results.
The THV instruments can also be delivered without measuring system. An electronic probe or a dial indicator shall thenbe used as measuring system.
Due to its small size, The THV can be moved easily. It is therefore frequently used as mobile calibration station.
IDEAL FOR THE USE IN A CLEAN ROOM AND IN THE
WORKSHOP AREA
THE INSTRUMENT MEETS THE REQUIREMENTS OF ALL
EN ISO 9000
VERY SIMPLE MANIPULATIONLARGE RANGE OF ACCESSORIESCHECKING OF INTERNAL AND EXTERNALDIMENSIONS USING A SINGLE MEASURING ELEMENTACCORDING TO THE APPLICATION, THE
INSTRUMENT CAN BE USED IN HORIZONTALPOSITION OR IT CAN BE INCLINED UPTO 90° USING
THE ADDITIONAL STAND

70

DESCRIPTION

Display units:

Heidenhain ND1100

PC with TRIMOS-WinDHI
software

Interchangeable inserts forexternal measurements

Knurled knob for the
displacement of the sliding
datum device

Interchangeable anvils forinternal measurements

Measuring force
adjustment

71

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

THV
display / software
Display unit Heidenhain ND1100

LINEAR MEASURING SYSTEM, MIN./MAX. VALUE HOLD
inversion of measuring direction sign (+/-)
ZERO SETTING OF THE DISPLAY AND PRESET INPUTPARAMETER SETTINGS AND CLASSIFICATIONCONFIGURATION OF THE DISPLAY USING EXTERNALCONTACT
RS232 DATA OUTPUT

PC with TRIMOS WinDHI

TRIMOS-WinDHI Software allows the performance of all required measuring functions and the connection to a gauge
inspection and management system.
INVERSION OF MEASURING DIRECTION SIGN (+/-)
DDE-SERVER (FOR EXCEL, WORD, ETC.)
GRAPHIC HELP FOR MEASURING FUNCTIONSDATA TRANSFER USING A FOOT PEDALDIRECT DISPLAY OF ALL LENGTH MEASURING VALUESAND MIN. / MAX. VALUE HOLD
MEASURING WITH 9 REFERENCESCOMPATIBLE WITH MEASURING EQUIPMENT
MANAGEMENT SYSTEMSTECHNICAL SPECIFICATIONS
THVAbsolute measuring range

mm (in)50 (2)

Application range (external measurements)

mm (in)100 (4)

Application range (internal measurements)

mm (in)100 (4)µm

0.2+ L (mm)/250

µm

0.1

Max. permissible errors
Repeatability (2s)1)1)

Resolutions (depending on display unit)

1)

mm (in)0.1 ÷ 0.00001 (.0001 ÷ .000001)

Measuring force (adjustable)

N

0÷4

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weightkg

22

Instruments with integrated measuring system. Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

72

SCHEMA

50÷100

157

172

232

99

110

447

STANDARD INSTRUMENTThe THV instruments are supplied as follows:
Instrument according to specifications (without display)
Measuring inserts for external measurements (TELS50)
Measuring anvils for internal measurements (THV-10)
Protection cover (THV.HO.0-50)
User’s manual (750 50 0006 03)
Test certificate

CODE NUMBERS
THV

Rigid table

Floating table

Instruments with measuring system

THVR.0-50 700 206 00 22

THVR.0-50S 700 206 00 24

Instruments without measuring system

THV.0-50 700 206 00 21

THV.0-50S 700 206 00 01
73

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

THV
Applications

Calibration of plain and threadedplug gauges using the floating table(THV-115)

Calibration of plain rings gauges(THV-260)

Measuring of precision parts(THV-100 / THV-101)

Checking of test indicators(THV-100 / THV-150

Adjustable support for a constantmeasuring force (THV-200)

Floating table for more accurateinternal measurements

74

LABCONCEPT+ Premium75

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT + Premium
introduction
The Labconcept and Labconcept Premium are high precisioncalibration systems that meet the most sophisticated requirements.
The up-to-date, well designed modular concept enables extremely secure functioning, facilitates the manipulation andtherefore increases the productivity in the measuring laboratory. Simplicity and high accuracy have been associated.
This new concept, integrating a computer and a touch screen
as well as the appropriate Trimos-WinDHI software with all necessary measuring functions guarantees the best results. A temperature compensation system as well as a gauge management
system can be installed to enhance the system performance.
Instruments with a measuring range from 300 to 2000 mm areavailable, all made in one single piece. All measuring ranges
being direct, it means that the whole measuring range isavailable without adjustment or intermediate re-calibration.
MEETS THE REQUIREMENTS OF ALL EN ISO 9000
PC WITH EXCLUSIVE SOFTWARE WINDHIHIGH PRECISION MEASURING SYSTEMDIMENSIONALLY STABLE INSTRUMENT BASE
ADJUSTABLE MEASURING FORCE (FROM 0 TO 12 N)
LARGE RANGE OF ACCESSORIESDIRECT MEASUREMENT OVER THE WHOLE
MEASURING RANGE

76

DESCRIPTION

PC with TRIMOS-WinDHI
software for all required
measuring functions

Interchangeable
accessories

Display of themeasuring force

Dimensionally stable instrument
base positioned on three level points(two additional balance points).

Carriage with locking andfine adjustment device

77TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT + Premium
display / software
TRIMOS WinDHI

TRIMOS® WinDHI Software allows the performance of all
required measuring functions. It can be connected to thetemperature compensation software WinComp and to any
gauge inspection and management program.
DDE-SERVER (FOR EXCEL, WORD, ETC.)
GRAPHIC HELP FOR MEASURING FUNCTIONSDATA TRANSFER USING A FOOT PEDALDIGITAL DISPLAY OF THE SELECTED MEASURING
FORCE IN NEWTON (N)DIRECT DISPLAY OF ALL LENGTH MEASURING VALUESAND MINI/MAX VALUE HOLD
INPUT OF 9 PRESET VALUESINVERSION OF MEASURING DIRECTION SIGN (+/-)
COMPATIBLE WITH TEMPERATURE COMPENSATIONSYSTEM TEMPCOMPQMSOFT

Trimos recommends the QMSOFT software package for inspection and management of all measuring tools available.
INTEGRATED DRIVERS FOR TRIMOS INSTRUMENTSREQUIRED NOMINAL SIZES AND TOLERANCESAVAILABLE ACCORDING TO ALL STANDARDSCUSTOMIZED INSPECTION CERTIFICATES

78

display / software
TEMPERATURE COMPENSATION SYSTEM TRIMOS TEMPCOMPThe Temperature Compensation System TempComp gives asolution to air conditioning problems in measuring laboratories.
COMPATIBLE WITH HPD, LABCONCEPT, LABCONCEPTPREMIUM AND LABCONCEPT NANO INSTRUMENTS
TRIMOS WINCOMP EXCLUSIVE SOFTWAREACQUISITION AND MANAGEMENT OFTEMPERATURE DATAPERMANENT CONNECTION WITH WIN DHI
REAL-TIME COMPENSATION OF THE MEASUREMENTTEMPERATURE EVOLUTION HISTORY OVER SEVERALYEARS FOR A PERFECT TRACEABILITY
GRAPHICAL DISPLAY OF TEMPERATURE EVOLUTION
MATERIALS LIBRARY
INDICATION OF THE MEASUREMENT RELIABILITY LEVEL

TempComp Basic

A: Part sensor

BASIC TEMPERATURE COMPENSATION SYSTEM
2 TEMPERATURE SENSORS:

B: Carriage
sensorA PART TO BE MEASUREDB MEASURING CARRIAGE

C: Instrument
sensor

TempComp PremiumPART TO BE MEASURED
3 TEMPERATURE SENSORS:
A PART TO BE MEASUREDB MEASURING CARRIAGEC INSTRUMENT BASE

TempComp Basic & Premium
Application range (temperature)°C+16 ÷ +24

Max. resolution (temperature)

°C

0.01

Max. permissible errors (temperature)

°C

0.05

79

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT + Premium
display / software
TempComp Advanced

The environnement control system TempComp Advanced
represents an evolution of the temperature compensation
system TempComp.
Tempcomp is an exclusive temperature compensation system with environmental parameter verification of the laboratory. The temperature acquisition is managed by WinComp
Advanced software. It offers, on top of WinComp functionalities, the possibility to check the laboratory in real time via
Internet, Intranet, mobile phone, etc..
INTEGRATED TEMPERATURE COMPENSATION SYSTEM
FOR LABORATORY
3 TEMPERATURE SENSORS ON THE INSTRUMENT:• PART TO BE MEASURED• MEASURING CARRIAGE• INSTRUMENT BASE
4 TEMPERATURE SENSORS IN THE LABORATORY
1 RELATIVE HUMIDITY SENSOR
1 ATMOSPHERIC PRESSURE SENSOR

TempComp Advanced
Application range (temperature)°C+16 ÷ +24

Max. resolution (temperature)

°C

0.01

Max. permissible errors (instrument temperature)

°C

0.05

Max. permissible errors (environmental temperature)

°C

0.16

Max. permissible errors (humidity)%±2

Max. permissible errors (pressure)mbar± 0.5 %

80

TECHNICAL SPECIFICATIONSLabconcept
Measuring range

1)

mm (in)

500

1000

1500

2000

550 (21)1050 (41)1550 (61)2050 (80)Max. permissible errors 1)µm

0.3 + L (mm) / 1500

Repeatability (2s) 1)

µm

0.1

Resolutions

mm (in)0.01 ÷ 0.00001(.0001 ÷ .000001)

Max. displacement speed

mm / s1500

Measuring forceN

0 ÷ 12

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Relative humidity%

20 ÷ 80

Weight

kg

94

152

181

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

Labconcept Premium
Measuring range

1)

123

300

500

1000

550 (21)mm (in)370 (14)

Max. permissible errors 1)µm

0.1 + L (mm) / 2000

Repeatability (2s) 1)

µm

0.05

Resolutions

mm (in)0.01 ÷ 0.00001 (.0001 ÷ .000001)

Max. displacement speed

mm / s400

Measuring forceN

0 ÷ 12

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Relative humidity%

Weightkg

1050 (41)0.15 + L (mm) / 2000

20 ÷ 80
78

95

125

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

427

SCHEMA515 / 616 / 892 / 1170 / 1446

270

942 / 1122 / 1622 / 2122 / 2622

81

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT + Premium
STANDARD INSTRUMENT
The Labconcept and Labconcept Premium instruments are supplied as follows:
Instrument according to specifications
Pair of anvils with tungsten carbide surface (HPA-1)
PC with interface, touch screen 1) with adjustable support 1) and touch screen pen 1)
Foot pedal for data transfer (TELMA31)
Opto-RS connection cable for measuring force (TVM.O-PC / AT.9P)
Lapping plate (TA-TO-302)
Protection cover (TEL.HO500 / 1000 / 1500 / 2000)
Allen key set (TA-TO-004)
User’s manual (750 50 0015 03)
Test certificate
1)

Not included in versions LABC-B. Touch screen replaced by a regular TFT screen.

CODE NUMBER
With touch screen

With TFTscreen

Labconcept

LABC500 700 203 10 01

LABC500B 700 203 10 02

Measuring range 500 mm

LABC1000 700 203 20 01

LABC1000B 700 203 20 02

Measuring range 1000 mm

LABC1500 700 203 30 01

LABC1500B 700 203 30 02

Measuring range 1500 mm

LABC2000 700 203 40 01

LABC2000B 700 203 40 02

Measuring range 2000 mm

With touch screen

With TFTscreen

Labconcept Premium

LABCP300B 700 203 10 13

Measuring range 300 mm

LABCP500 700 203 10 11

LABCP500B 700 203 10 12

Measuring range 500 mm

LABCP1000 700 203 20 11

LABCP1000B 700 203 20 12

Measuring range 1000 mm

82

APPLICATIONSCalibration of ring gauges(TA-SU-313 / TEL16.1 / HPA-1)

Calibration of plug gauges(HPA-1 / TULM6 / L05 / LABC-15)

Calibration of small ring gauges(TA-SU-313 / LABC-70 / TA-SU-354)

Calibration of thread ring gauges(TA-SU-313 / LABC-70 / TA-SU-354)

Calibration of thread pluggauges (HPA-1 / TEL6 / 3P / 0.173.2 / S6.5 / LABC-15)

Calibration of external micrometers(HPA-1 / TULM14)

Comparative checking of gauge
blocks < 250 mm (TA-SU-313 / TASU-305)

Comparative checking of gauge blocks> 250 mm (TA-SU-313 / TELMA7 / P / TASU-305 / TA-SU-306)

Setting of 2-point internal micrometers(HPA-1 / TELMA7 / TELMN7.2)

83

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT + Premium
APPLICATIONSChecking of dial indicators (TULM5C)

Checking of test indicators (TULM15)

Checking of snap gauges(TA-SU-313 / TEL14N)

Temperatue compensation system
TempComp

Checking of taper thread ring gauges(HPA-1 / LABC80)

Special design for measurement oflarge taper threaded gauges

Direct measurement over the whole measuring range.

84

LABCONCEPT
Nano

85

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT Nano
introduction
No compromise on accuracy
The Labconcept Nano is a new reference in the field of dimensional metrology. It integrates 40 years of knowledge
and continuous improvement. It is a remarkable instrument
for all measuring tasks that require extremely high accuracy.
The uncompromising design of the Labconcept Nano offers
an ideal and performing platform for checking and calibrationof all kinds of gauges and measuring instruments. Checking
of internal and external dimensions will be done as a fully
automatic procedure by the three motorized axes XYZ andthe legendary simplicity of use of Trimos WinDHI software.
The Labconcept Nano is completely designed and manufactured in Switzerland according to the highest quality standards.
Robustness, reliability and longevity are our traditional values.
A New Technological Dimension
The Labconcept Nano combines tradition, experience anda strong technological lead. It integrates the latest measuring and motorisation technologies and can be consideredas the first “full digital” calibration instrument. A regular PC
controls all electronic components. This low-power solution avoids heating and keeps the energy, maintenance andrepair costs at a reasonable level. The linear bearings usedin all guideways have proven their superiority over all othertechnologies in terms of precision, wear, rigidity, temperaturestability, reliability, dust protection and maintenance. They
ensure exceptional repeatability and precision through time.

86

UNEQUALLED HIGH LEVEL OF ACCURACY
EXCEPTIONAL REPEATABILITY
MOTORIZED MEASURING CARRIAGE, X AXIS,SELECTION OF SPEED BY SOFTWARE
MOTORIZED UNIVERSAL MEASURING TABLE, CNC Y
AND Z AXIS WITH INTEGRATED MEASURING SYSTEMMEASURING FORCE (0-12N) AND LOCKING OF
MEASURING ANVIL PERFORMED BY SOFTWAREINTEGRATED TEMPERATURE COMPENSATION SYSTEM
ABSOLUTE MEASURING RANGE ON
ALL MODELS : 350 MM
APPLICATION RANGES OF 350, 600 AND 1100 MM
MEASURING OF PARTS UP TO 60 KG IN WEIGHT
2 SCREENS IN THE STANDARD PACKAGE
CNC CONTROLLED MEASUREMENTS EVEN ON
DIAMETERS AND THREADSDESCRIPTION

motorized measuringtable (YZ)

CNC controlled
measuring carriage (X)PC with 2 screens including
WinDHI software formeasurements and
WinComp for temperaturecompensation

Ergonomic andcompact Workbench

87

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT Nano
display / software
TRIMOS WinDHI Nano

Trimos Win DHI Nano is the exclusive measurement softwareof Trimos. It is part of the basic equipment of the Labconcept
Nano and allows the performance of all measuring functions.
It helps the user to perform all measurement tasks through auser-friendly interface.
The motorization of the measuring carriage (X) and both vertical (Z) and horizontal (Y) axis of the universal measuring table allow an exceptional performance in terms of measuring
speed, ease of use and accuracy.
Positioning can be done easily using the mouse and the keyboard or via the touch screen (optional) or a joystick(optional). Once positioned, measurements are entirely CNC
controlled, including searching the reversal point. Plug andring gauges, threaded ring and plug gauges etc. can be measured automatically in a few seconds. No risk of damage tothe sensitive probes while moving or measuring, even withtiny parts and inserts.
100 % AUTOMATIC MEASUREMENTS
MEASUREMENTS PERFORMED IN A FEW SECONDS
USER FRIENDLY INTERFACE
GRAPHIC HELP FOR MEASURING FUNCTIONSELECTRONICALLY ADJUSTABLE MEASURING FORCEDATA TRANSFER USING A FOOT SWITCHDDE SERVER (FOR EXCEL, WORD, ETC.)

TRIMOS WINCOMP

The Labconcept Nano is equipped as standard with a temperature compensation system Trimos WinComp allowing
the acquisition and management of temperature data.
TRIMOS WINCOMP EXCLUSIVE SOFTWAREACQUISITION AND MANAGEMENT OFTEMPERATURE DATAPERMANENT CONNECTION WITH WIN DHI
REAL - TIME COMPENSATION OF THE MEASUREMENTTEMPERATURE EVOLUTION HISTORY OVER SEVERALYEARS FOR A PERFECT TRACEABILITY
GRAPHICAL DISPLAY OF TEMPERATURE EVOLUTION
MATERIALS LIBRARY
INDICATION OF THE MEASUREMENT RELIABILITY LEVEL

88

display / software
QMSOFT

Trimos recommends the QMSOFT software package for inspection and management of all measuring tools available.
INTEGRATED DRIVERS FOR TRIMOS INSTRUMENTSREQUIRED NOMINAL SIZES AND TOLERANCESAVAILABLE ACCORDING TO ALL STANDARDSCUSTOMIZED INSPECTION CERTIFICATESTECHNICAL SPECIFICATIONSLabconcept Nano

600

1100

350 (13.2)

600 (23.6)

1100 (43.3)

Application range

mm (in)Absolute measuring range

mm (in)350 (13.2)

Max. permissible errors 1)µm

0.07+L(mm) / 2000

Repeatability (2s) 1)

µm

0.03

mm (in)0.000001 (0.0000001)

Max. resolution

1)

350

Measuring force (adjustable by software)N

0 ÷ 12

Operational temperature°C+15 ÷ +35

Temperature of storage°C-10 ÷ +40

Relative humidity%

20 ÷ 80

Weight

kg

350

420

500

Values valid at temperature of 20 ± 0.2 °C and relative humidity of 50 ± 5%.

Measuring table with motorized Y and Z axes
Z axis, displacement range 2)

mm (in)100 (4)Y axis, displacement range

2)

mm (in)50 (2)

X axis, floating movement

µm

± 10

Angle of inclination (Y)

°

± 1.5

Angle of rotation (Z)°±4Max. weight of parts

kg

60

2)

Both axes Y and Z have an integrated measuring system.

89

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT Nano
SCHEMA

STANDARD INSTRUMENT
The Labconcept Nano are supplied as follows :Instrument according to specifications with tungsten carbide surface anvils
Universal measuring table with motorized Y and Z axes (Nano-14)
PC with interface, 2 LCD TFT screens and 1 printerFoot pedal for data transfer (TELMA31)
Specially designed workbench with 1 drawer cabinet and 1 door cabinet
Temperature compensation system (TEMPCOMP-B)
Lapping plate (TA-TO-302)
Protection cover (TEL.HO500 / 1000 / 1500)
Allen key set (TA-TO-004)
User’s manual (750 50 0039 03)
Test certificate

CODE NUMBERS
Labconcept Nano

Standard workbenchAnti-vibration workbench

LABC-NANO 350 700 213 00 01

TA-TO-306 714 12 006

TA-TO-310 714 12 010

Measuring range 350 mm

LABC-NANO 600 700 213 10 01

TA-TO-307 714 12 007

TA-TO-311 714 12 011

Measuring range 600 mm

LABC-NANO 1100 700 213 20 01

TA-TO-308 714 12 008

TA-TO-312 714 12 012

Measuring range 1100 mm

90

APPLICATIONSCalibration of plain ring gauges(TEL16.1 / TA-SU-354)

Calibration of small plain ring gauges(TA-MS-370 / TEL76 / TA-SU-354)

Calibration of thread ring gauges(TA-MS-370 / TEL75 / TA-SU-354)

Calibration of plug gauges(TULM6 / L05 / TA-SU-315)

Calibration of thread plug gauges(3P / 0.17-3.2 / S6.5 / TA-SU-315)

Comparative checking of gauge blocks> 250 mm (TA-SU-307 / TEL7 / TELMA7)

Calibration of gauge bars
(TELMA7 / TELMN7.2)

Checking of taper thread ring gauges(TA-MS-381 / TEL75)

Checking of taper thread plug gauges(TA-MS-381 / TEL75)

91

TRIMOS | HORIZONTAL MEASURING INSTRUMENTSLABCONCEPT Nano
APPLICATIONSCalibration of snap gauges(TEL16.1 / TEL14N)

CNC-controlled measurement with
automatic reversal point search.

Anti-vibration table (optional)

Integrated temperature compensation
system TempComp

CNC controlled internal taper ringmeasurement

CNC controlled external taper ringmeasurement

CNC controlled internal measurementof a bearing ring

CNC controlled externalmeasurement of a bearing ring

Calibration of a specific gauge

92TELS50
Ø6,5

Ø8

Ø6,5

TELS5240

LABCN

LABC + P

THV

HPD

HPA

HG

•

Pair of measuring anvils Ø 8 mm•

• • • • • •Pair of measuring anvils Ø 2 mm•

• • • • • •Pair of measuring anvils Ø 1 mm•

• • • • • •Pair of a knife-shaped anvils••

Ball probe Ø 1.00 mm••

Ball probe Ø 1.250 mm••

Ball probe Ø 1.50 mm••

Ball probe Ø 1.75 mm••

Ball probe Ø 2.00 mm••

Ball probe Ø 2.032 mm••

Ball probe Ø 2.20 mm••

Ball probe Ø 2.25 mm••

Ball probe Ø 2.50 mm••

Ball probe Ø 2.75 mm••

Ball probe Ø 3.00 mm••

Ball probe Ø 3.20 mm••

Ball probe Ø 3.25 mm••

Ball probe Ø 3.50 mm••

609 05 047

TELS53

H

Pair of measuring anvils Ø 6.5 mm

609 05 044

42

ALESTATELSACCESSOries

TELS53/D1

42

SP609 05 048 01

Ø6,5

H

Ø6,5

Ø

609 05 048•

TELS50-L05
SP609 05 044 01

42

TA-MI-301
279 901008 001

TA-MI-302
279 901008 002

TA-MI-303
279 901008 003

TA-MI-304
279 901008 004

TA-MI-305
279 901008 005

TA-MI-306
279 901008 006
Ø3,5TA-MI-307Ø

279 901008 007
15,5

TA-MI-308
279 901008 008

TA-MI-309
279 901008 009

TA-MI-310
279 901008 010

TA-MI-311
279 901008 011

TA-MI-312
279 901008 012

TA-MI-313
279 901008 013

TA-MI-314
279 901008 014

93TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

TA-MI-315

Ø

Ø3,5

15,5

•

Ball probe Ø 4.50 mm••

Ball probe Ø 5.00 mm••

Ball probe Ø 5.50 mm••

Ball probe Ø 6.00 mm••

Ball probe Ø 6.50 mm••

Ball probe Ø 7.00 mm••

Ball probe Ø 8.00 mm••

Ball probe Ø 9.00 mm••

Ball probe Ø 10.00 mm••

279 901008 020

TA-MI-321
279 901008 021

TA-MI-322
279 901008 022

TA-MI-323
279 901008 023

TA-MI-324
279 901008 024

TA-MI-325
279 901008 025

160

HPA-1
Ø12

33

Pair of standard anvils

609 05 017

TA-MI-350
609 05 105Pair of anvils with radius 30 mm forgauge block

TEL1

Anvil with tungsten carbide surface

Ø20

Ø12

89

609 05 040

50

TEL6

Ø

TEL6/6609 05 079

94

• •

•

LABCN

LABC + P••• •Pair of anvils Ø 4 mm

• • • •• •Pair of anvils Ø 6 mm

• • • •• •Pair of anvils Ø 6.35 mm• • • •

• •609 05 078

TEL6/6.35• •• • • •

609 05 077
86•

Pair of anvils Ø 6.50 mm

609 05 041

TEL6/4THV•

279 901008 019

TA-MI-320

HPD

Ball probe Ø 4.00 mm

279 901008 018

TA-MI-319

HPA•

279 901008 017

TA-MI-318

HG•

279 901008 016

TA-MI-317

H

Ball probe Ø 3.70 mm

279 901008 015

TA-MI-316

ALESTATELSACCESSOries

TEL6/6.8

Ø

LABCN

LABC + P

THV

HPD

HPA

• •

Pair of anvils Ø 8 mm

• • • •• •Pair of anvils Ø 10 mm• • • •• •Pair of a knife-shaped anvils• • • •• •Pair of anvils with ball Ø 10 mm• • • •• •Pair of parallel attachments 25 mm• • • ••

Pair of parallel attachments 1"• • • ••

Pair of parallel attachments 5 mm,
for ext. measurement from 20 mm

•

• •

•

Pair of parallel attachments .2",
for ext. measurement from .8"

•

• •

•

Pair of holders with measuring inserts
TELS50, H=40 mm

•

• •

•

TELMA5.0/H70 Pair of holders with measuring inserts
SP609 05 005 01
TELS50, H=70 mm

•

• •

•

TELMA5.0/H88 Pair of holders with measuring inserts
SP609 05 005 02
TELS50, H=88 mm

•

• •

•

TELMA5.0/H55 Pair of holders with measuring inserts
SP609 05 005 03
TELS50, H=55 mm

•

• •

•

TELMA5.0/H65 Pair of holders with measuring inserts
SP609 05 005 04
TELS50, H=65mm

•

• •

•

609 05 081

TEL6/10609 05 079

0,5

HG

• • • •

TEL6/890,5HPair of anvils Ø 6.8 mm

609 05 080

86

ALESTATELSACCESSOries

TULM6/L05
609 05 016

TEL7
609 05 013
52

56

35

TEL5
609 05 101

TEL5E

121

609 05 102

30

15

TEL5.10612 12 002

TEL5.10E125

612 12 007

TELMA5.0
609 05 005

171

H

51

TELS10

15

Pair of inserts for int. meas.
>Ø10, H=20 mm•

Pair of inserts for int. meas.
>Ø10, H=50 mm•

SP609 05 043 01

Pair of inserts for int. meas.
>Ø10, H=35 mm•

TELS10/1.5C

Pair of inserts for int. meas. >Ø3 mm•

H

609 05 042

TELS10/H50
609 05 043

20

5TELS10/H35

SP609 05 070 01
15

95

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

HPair of inserts for int. meas.
>Ø6, H=28 mm•

Pair of inserts for int. meas.
>Ø13, H=26 mm•

609 05 060

Pair of inserts for int. meas.
>Ø11, H=30 mm•

TELS13

Pair of measuring inserts flat-ball•

TELS11H23

609 05 058

TELS11.1
609 05 103

28

H

15

TELS12LABCN

LABC + P

THV

HPD

HPA

HG

H

15

ALESTATELSACCESSOries30

15609 05 033

26

20

THV-10609 05 034

Pair of inserts for internal meas.
>Ø10 mm, (rigid measuring table)•

Pair of inserts for internal meas.
>Ø2.5 mm, (rigid measuring table)•

Pair of inserts for internal meas.
>Ø13 mm, (rigid measuring table)•

Pair of inserts for internal meas.
>Ø5 mm, (rigid measuring table)•

Paire de touches mesures int.
>Ø10 mm, (rigid measuring table)•

Pair of inserts for internal meas.
>Ø2.5 mm, (floating measuring table)•

Pair of inserts for internal meas.
>Ø13 mm, (floating measuring table)•

Pair of inserts for internal meas.
>Ø5 mm, (floating measuring table)•

Pair of inserts for internal meas.
>Ø10 mm, (floating measuring table)•

26

THV-11

37

20

THV-12609 05 032

20

609 05 035

THV-152228

609 05 092

THV-20609 05 037

26

20

THV-21609 05 038

44,5

20

THV-22609 05 039

36

22

THV-25609 05 091

24

36

THV-26609 05 094

H

TEL16.1
L96

609 05 088

Pair of L-shaped probes,
H=25 mm, L=50 mm

• •

• •

TEL16.2

LABCN

LABC + P

THV

HPD

HPA

HG

H

Pair of L-shaped probes,
H=40 mm, L=55 mm

• •

• •

Pair of L-shaped probes,
H=70/45 mm, L=65mm

• •

• •

H

609 05 089

ALESTATELSACCESSOries

TEL16.3

L609 05 097

TEL9
612 12 008

Pair of inserts for internal meas.
>Ø20 mm, with air cushion• •Pair of inserts for internal meas.
>Ø10 mm, with air cushion• •Pair of inserts for internal meas.
>Ø10 mm, without air cushion• •Pair of measuring devices for
snap-gauges• • • •

120

30

TEL9.10612 12 009

TELMN9612 11 002

TELMA8612 12 052

TEL75.01
298 000205 050

TEL75.1
298 000205 051

TEL75.2
298 000205 052

TEL75.3
298 000205 053

TEL75.4
298 000205 054
Ø

TEL75.5
298 000205 055

TEL75.6
298 000205 056

TEL75.7
298 000205 057

TEL75.8
298 000205 058

TEL75.9298 000205 059

TEL75.10
298 000205 060

TEL75.11
298 000205 061

TEL75.12
298 000205 062•

T-shaped measuring insert, ruby balls
Ø0.335 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø0.455 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø0.530 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø0.620 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø0.725 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø0.895 mm

• •

• •

T-shaped measuring insert, ruby ballsØ1.100 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø1.350 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø1.650 mm

• •

• •

T-shaped measuring insert, ruby ballsØ2.050 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø2.550 mm

• •

• •

T-shaped measuring insert, ruby ballsØ3.200 mm

• •

• •

T-shaped measuring insert, ruby balls
Ø4.000 mm

• •

• •

97

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

TEL76.2
509 05 20 0018

TEL76.3
M2,5

Ø

509 05 20 0019

TEL76.4
509 05 20 0020

TEL76.5509 05 20 0021

TEL77.100
279 901009 001

TEL77.125
279 901009 008

TEL77.150
279 901009 002

TEL77.175
279 901009 009

TEL77.200
279 901009 003

TEL77.250
279 901009 004

TEL77.300
TEL77.350
ØM3

279 901009 005

279 901009 006

TEL77.400
279 901009 007

TEL77.450
279 901009 010

TEL77.500
279 901009 011

TEL77.550
279 901009 012

TEL77.600
279 901009 013

TEL77.700
279 901009 014

TEL77.900
279 901009 015

TEL77.1000
279 901009 016

98

LABCN

LABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries

Measuring insert with ruby ball
Ø0.5 mm• •• •Measuring insert with ruby ball
Ø1.0 mm• •• •Measuring insert with ruby ball
Ø2.5 mm• •• •Measuring insert with ruby ball
Ø5.0 mm• •• •Pair of measuring inserts with ball
Ø1.00 mm

• •• •Pair of measuring inserts with ball
Ø1.25 mm

• •• •Pair of measuring inserts with ball
Ø1.50 mm

• •• •Pair of measuring inserts with ball
Ø1.75 mm

• •• •Pair of measuring inserts with ball
Ø2.00 mm

• •• •Pair of measuring inserts with ball
Ø2.50 mm

• •• •Pair of measuring inserts with ball
Ø3.00 mm

• •• •Pair of measuring inserts with ball
Ø3.50 mm

• •• •Pair of measuring inserts with ball
Ø4.00 mm

• •• •Pair of measuring inserts with ball
Ø4.50 mm

• •• •Pair of measuring inserts with ball
Ø5.00 mm

• •• •Pair of measuring inserts with ball
Ø5.50 mm

• •• •Pair of measuring inserts with ball
Ø6.00 mm

• •• •Pair of measuring inserts with ball
Ø7.00 mm

• •• •Pair of measuring inserts with ball
Ø9.00 mm

• •• •Pair of measuring inserts with ball
Ø10.00 mm

• •

• •

TELS51
Ø6,5

TELS51E

609 05 045

TELS18612 11 031
60

Holder for measuring inserts 4-48••

Pair of holders for measuring inserts
Ø 3.5 mm••

Pair of holders for measuring inserts
Ø 3.5 mm•LABCNLABC + P

THV

HPD

HPA

•

Pair of holders for measuring inserts
Ø 3.5 mm• • • •• •Pair of holders for measuring inserts
Ø 6.5 mm• • • •

• •609 00 019

Pair of cone holder for int. threadmeasurement• • • •TEL16.2SP2

Pair of L-shaped holders for M3 Inserts

TEL15
609 05 098

TEL5.0ØHG•

509 05 20 0028

TELS51/D3.5

H

Holder for measuring inserts M2.5

509 05 20 0027

42

ALESTATELSACCESSOries

612 11 017

147

40
M3

56

TEL18

• •

SP609 05 089 02

• •

55

Electronic probe 25 mm••

Electronic probe P25 withmagnetic insert••

276 950001 002

Electronic probe P25 with magnetic
insert, and certificate••

P25TA.1

Magnetic insert for probe P25

•

•

P25
276 950000 001

P25TA
276 950001 001

P25TA.C279 950001 001

TA-MS-301•

Electronic probe 30 mm

276 940001 004

TA-MS-302
276 000230 001

TEL3C.1
276 000230 002

TEL70706 200 14

TULM70706 200 13

LABC-70706 203 01•

Digital indicator 50 mm,
resolution 0.001 mm•• • • • • •Digital indicator 50 mm,
resolution 0.01 mm•• • • •

Electronic probe for internalmeasurement
Electronic probe for internalmeasurement
Electronic probe for internal measurement, (with support, without interface)

•

•
•
•

•
99

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

TA-MS-370
706 203 02

LABC80709 60 001

LABC80.1
712 04 003

TA-MS-381••

Measuring system for taper threads,
without electronic probe LABC-70.1••

Measuring system for cylindrical parts•

Pair of V-shaped supports TA-MS-303•

609 00 035

612 32 003

Plate

•

Plate with crossed V-grooves•

Plate with H-shaped clearance•

Support for plates•

602 33 004

602 33 003

TELS5.3
602 33 001

TELS5
612 20 001

THV-101

Plate with H-shaped clearance•

Holder for dial test indicators•

Support for THV-101 + THV-150•

602 33 002

THV-150609 00 005

THV-100
602 15 001

TELMA7245

175

602 39 005

100

Pair of vertically adjustable supports

•

• •

• •

Vertically adjustable support

•

• •

• •

602 39 007

TELMA7/P

LABCN•

TA-MS-303

TELS5.2

LABC + P

Measuring system for taper threads,
with electronic probe LABC-70.1

Measuring system for taper threads,
without electronic probe LABC-70.1

TELS5.1

THV•

Electronic probe for internal measurement, (with support, without interface)712 04 010

TA-SU-303

HPD

HPA

HG

H

ALESTATELSACCESSOries

TELMA7/SP04
602 39 012

TELMA7/SP05

245

175

602 39 013

HG-3

Pair of vertically adjustable supports,
Adjustment range Z=240 mm,
standard plate
Pair of vertically adjustable supports,
Adjustment range Z=240 mm,
U-shaped Plate

TA-SU-313
709 40 044

209

H-13
709 40 018

HG-13
709 40 016

464

210,5

HPA-14

• •

• •

•

• •

•

•

Universal measuring table <12 kg,
Positioning plate 250 x 85 mm

H-6

Support for plug gauges

144

603 00 007

THV-115709 40 035

THV-115.1
609 40 035

TA-SU-316
709 40 041

TA-SU-317
609 40 041•

Universal measuring table <12 kg,
Positioning plate 250 x 85 mm

224,5

LABCN

• •

Vertically adjustable support

Universal measuring table <40 Kg,
Positioning plate 360 x 120 mm

LABC + P••

709 40 033

THV

• •

Pair of vertically adjustable supports

Universal measuring table <12 kg,
Positioning plate 250 x 85 mm

HPD

• •

609 40 032
345

HPA

•

709 40 032

HG-3/P

HG

H

ALESTATELSACCESSOries••

Measuring table for plug gauges <Ø60
x 160 mm, with self centring clamp•

Measuring table for plug gauges <Ø60
x 160 mm, without self centring clamp•

Measuring table for plug gauges<Ø60 x 160 mm, with self centring
clamp, not for TA-SU-313

• •

• •

Measuring table for plug gauges<Ø60 x 160 mm, without self centring
clamp, not for TA-SU-313

• •

• •

96

74

THV-115.2
706 04 002

LABC-15709 40 034

LABC-15.1
609 40 034•

Self centring clamp, for THV-115.1 +TA-SU-317
Measuring table for plug gauges<Ø100 x 280 mm, with self centring
clamp and alignment screws• ••Measuring table for plug gauges<Ø100 x 280 mm, without self centring
clamp and alignment screws

• •

•

101

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

TA-SU-315
709 40 040

Nano-15.1
609 40 040

H-15
709 40 036

H-15.1
609 40 036

HG-15
709 40 037

HG-15.1
609 40 037

LABCN

LABC + P

THV

HPD

HPA

HG

•

Measuring table for plug gauges<Ø100 x 280 mm, without self centring
clamp and alignment screws•

Measuring table for plug gauges<Ø100 x 280 mm, with self centring
clamp and alignment screws•

Measuring table for plug gauges<Ø100 x 280 mm, without self centring
clamp and alignment screws•

Measuring table for plug gauges<Ø100 x 280 mm, with self centring
clamp and alignment screws•

Measuring table for plug gauges<Ø100 x 280 mm, without self centring
clamp and alignment screws•

Self centring clamp• • • •

• •708 03 007

Alignment screws for LABC-15.1,
Nano 15.1, H15.1, HG 15.1

• • • •

• •H-5

Pair of floating tables•

Pair of L-shaped supports•

Pair of V-shaped supports• • • •• •Pair of V-shaped supports•• •Supporting device for rolls• • • •

LABC-15.2
136

H

Measuring table for plug gauges<Ø100 x 280 mm, with self centring
clamp and alignment screws

92

706 04 001

LABC-15.3

606 32 001

TELMN7.1
612 32 002

TELMN7.2
612 32 001

TEL11
612 12 032

TELMN7.2/
SP01
612 23 001

102

ALESTATELSACCESSOries

• •

• •

•

•

23

69

Ø23

69

Ø23

69

TELMN7.2/
SP02

V-shaped roll for cylindrical parts• • • ••

Roll for Easy-Metric gauges• • • ••

Cylindrical roll• • • ••

Supporting device for 2 points bore
gauges• • • ••

Pair of supporting plates for 2 points
bore gauges <330 mm• • • ••

Pair of supporting plates for 2 points
bore gauges <180 mm• • • ••LABCN

LABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries

613 07 006

TELMN7.2/
SP03
613 07 007

TELMN7.2/
SP04
613 07 008

TA-SU-301
612 04 005

TEL17
602 32 014

TEL17.2
602 32 016

TEL17.1
602 32 015

TELMN4602 20 013

TA-SU-305
709 60 003

TA-SU-306
612 04 008

TA-SU-307
709 60 004

TULM14

Supporting device for 2 points bore
gauges• • • ••

Supporting device for gauge blocks
100-250 mm with anvils TA-MI-350• ••Additional support for gauge blocks>250 mm, to be used with TA-SU-305,(to be mounted on TELMA7)• ••Adjustable support for setting ofreversal point

Supporting device for gauge blocks(2 items): 1 support for gauge blocks
100-250 mm, 1 additional support forgauge blocks >250 mm

708 209 140

Clamping device for external
micrometers,12-20 mm/.48"-4"

TULM19Holding device for depth calipers

612 30 003

•

•

• •

•

• • • ••

103

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

76

LABCN

LABC + P

THV

HPD

• •

612 02 007

Pair of holders for dial indicators
Ø 3/8"•• ••TULM5C

Holder for dial indicators Ø 8 mm

• •

• •

Holder for dial indicators Ø 3/8"

• •

• •

Holding device for dial test indicators• ••Clamping device for snap gauges• ••Alignment device for snap gauges• ••Pair of ball attachments Ø10 mm• • • ••

Pair of ball attachments Ø1/2 "• • • ••TEL5CNE

612 02 003

80

HPA

•

612 02 008

TULM5CE612 02 002

TULM15609 02 015

TEL14N612 12 049

TA-SU-302
603 00 021

TEL10

58,5

HG

Pair of holders for dial indicators
Ø 8 mmTEL5CN

251,5

H

123

ALESTATELSACCESSOriesØ

612 12 023
35

TEL10E612 12 022

709 60 002

Pair of holders for large size
micrometers• ••LABC20.1

Holder for large size micrometers• ••LABC20175

289

609 60 002

THV-260
609 00 007

THV-261609 00 006

TA-SU-304
SP609 60 001 01
252

THV-180•

System for search of reversal point
for ring gauges (ANSI/ASME)•

Supporting device for large ringgauges, (only for use with HPA-14)

• •

• •

Micrometer spindle for checking ofdial indicators•

Pneumatic system for measuringforce setting•

4

606 25 008

System for search of reversal point
for ring gauges (DIN/ISO)

THV-500
616 00 021

104LABCNLABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries75

TEL14.1
H

606 00 004

TEL14.2606 00 005

TA-SU-354
706 02 004

TA-SU-355H

706 02 005

TA-SU-356
706 02 006

TA-SU-357
706 02 007

TA-SU-358
706 02 008

According to type

TA-SU-451
279 906001 028

TA-SU-407
279 906001 039

TA-SU-410
279 906001 073

TA-SU-411
279 906001 074

TA-SU-412
279 906001 075

TA-SU-419
279 906001 044

TA-SU-420
279 906001 069

TA-SU-483
279 906001 091

TA-SU-484
279 906001 092

Pair of clamping attachments
H<40 mm• • • •• •Pair of clamping attachments
H<60 mm• • • •• •Pair of fast clamping attachments
H<60 mm• • • •• •Pair of fast clamping attachments
H<100 mm• • • •• •Pair of fast clamping attachments
H<150 mm• • • •• •Pair of fast clamping attachments
H<200 mm• • • •• •Pair of fast clamping attachments
H<250 mm• • • •• •Specific adapter for 2-point bore
gaugesAdapter set for external micrometers••

Adapter for 2-points bore gauges
without ring Ø8-10 or 14 mm•

Fixing bush Ø8 mm for
TA-SU-407/8/9•

Fixing bush Ø10 mm for
TA-SU-407/8/9•

Fixing bush Ø14 mm for
TA-SU-407/8/9•

Adapter for 3 points internal
micrometers, Ø 25-140 mm•

Adapter for 3 points internal
micrometers, Ø 125-240 mm•

Adapter for 3 points internal
micrometers, Ø 245 - 385 mm•

Adapter for 3 points internal
micrometers, Ø >385 mm•

105TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

Universal adapter for 2-points bore
gauges, (to be mounted on mobile
probe)•

Universal dapter for 2-points bore
gauges, Ø 35-180 mm•

Universal dapter for 2-points bore
gauges, Ø 80-330 mm•

Universal dapter for 2-points bore
gauges, Ø 120-170 mm•

Universal dapter for 2-points bore
gauges, Ø 170-220 mm•

Universal dapter for 2-points bore
gauges, Ø 160-600 mm•

Adapter for external micrometers, andext. groove instruments Kroeplin•

Pair of adapter for internal/external
calipers (pair)•

Adapter for int. groove instruments
Kroeplin•

279 906001 066

Adapter for 2 points internal
micrometers•

TA-SU-406

Support for left bore gauges•

Adapter for dial indicators•

Adapter for test indicators•

TA-SU-401
279 906001 051

TA-SU-402
279 906001 052

TA-SU-403
279 906001 053

TA-SU-405
279 906001 062

TA-SU-423
279 906001 089

TA-SU-404
279 906001 054

TA-SU-414
279 906001 063

TA-SU-417
279 906001 064

TA-SU-416
279 906001 065

TA-SU-418

279 906001 070

TA-SU-421
279 906001 071

TA-SU-422
279 906001 072

M2.5

25
Ø5
Ø

THV-181
601 01 002

Ø20

106

TELS3.1C

Extension for dial indicator, 25 mm

279 918101 006

102
124,3

THV-181E
601 01 001•

Holder for checking dial indicators,
Ø 8mm

•

Holder for checking dial indicators,
Ø 3/8"•LABCN

LABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries

L

LABCN

Extension for ball and T-shaped
probes, L=15 mm•• •TEL76.1

Holder for ball and T-shaped probes•• •512 12 20 0003

TELS3C

Clamping sleeve for dial indicators
Ø 8 mm•Clamping sleeve for dial indicators
Ø3/8"

•

12

603 12 002
Ø

LABC + P

501 04 20 0002

Ø4
Ø3

Ø8,5

THV

• •

TEL76.7

6

HPD•

501 04 20 0001

8

HPA

Extension for ball and T-shaped
probes, L=7.5 mm

TEL76.6

26

HG

H

ALESTATELSACCESSOries

TELS3CE603 12 001

ØClamping sleeve for dial indicators
Ø 8 mm• • • ••

Clamping sleeve for dial indicators
Ø3/8"• • • ••

Dial indicator holder Ø8 mm for
TA-SU-313, H-13, HG-13• • • ••

612 02 013

Dial indicator holder Ø3/8 " for
TA-SU-313, H-13, HG-13• • • ••

H-32

Printer holder for Horizon

•

37,5

TEL3.1603 12 006

TEL3.1E
603 12 007

40
100

TULM13.2
612 02 012

TULM13.2E

612 06 004
260

Support for display unit Heidenhain
ND287

•

•

612 05 003

Support for display unit Heidenhain
ND1100••

THV-200

Adjustable support

379

TULM30.1
612 05 002

THV30.1
280

295

335

Table for horizontal instruments,
Measuring range = 500 mm

•

• •

•

LABC-TAB1000 Table for horizontal instruments,
714 12 002
Measuring range = 1000 mm

•

• •

•

LABC-TAB1500 Table for horizontal instruments,
714 12 003
Measuring range = 1500 + 2000 mm

•

• •

•

1177

LABC-TAB500
714 12 001

1600

1177

882,5

882,5

1177

2100

2700

•

712 12 038

882,5

107

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

602 13 004

602 13 005

H

602 13 006

602 13 007

LABCN

LABC + P

THV•

TELA05A/H300 Support for HG6000, H=300 mm
602 13 008•

TELA05A/H200 Support for HG7000, 8000 +10000,
602 13 009
H=125 mm• • • • • • •279 901004 101

Set of thread measuring wires withholders, Pitches 0.25-5 mm, holder
bore Ø6.5 mm

TEL15/150

Set of probes for threads ISO 60°• • •Set of probes for threads
Whitworth 55°• • •Set of cones for internal threadmeasurement 60°• • •Set of cones for internal threadmeasurement 55°• • •605 01 012

TEL15/152
605 01 013

TEL18/50S
605 01 010

TEL18/50ES
605 01 011

Set of T-shaped measuring insertswith ruby balls

• • •

• •

605 01 004

Set of measuring inserts with ruby ball,for internal diameters >1 mm

• • •

• •

TEL25D4M

Setting ring cones 60°• • •Setting ring cones 55°

• • •

TEL75
605 01 009

TEL76

609 01 002

TEL25D4E
609 01 001

709 201 11

Master ring gauge Ø 40 mm withSCS certificate•• • • • • • •TULM40INT.
METAS

Master ring gauge Ø 40 mm with
Metas certificate•• • • • • • •TULM40D.INT

709 201 31

108

HPD

•

TELA05A/H400 Support for HG5000, H=400 mm

3P/0.17-3.2/
S6.5

HPA

•

TELA05A/H500 Support for HG4000, H=500 mm

300

HG

•

TELA05A/H600 Support for HG3000, H=600 mm

390

H

•

TELA05A/H700 Support for HG2000, H=700 mm
303

ALESTATELSACCESSOriesLABCNLABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries709 201 01

Master plug gauge Ø 50 mm withSCS certificate• • • •• •TULM50EXT.
METAS

Master plug gauge Ø 50 mm with
Metas certificate• • • •• •Allen key set• • • •• •Allen key 1.5 mm• • • •

• •Lapping plate• • • •• •TULM50D.EXT

709 201 21

TA-TO-004
290 000911 100

TA-TO-003
290 000911 003

TA-TO-302
514 02 20 0002

TA-TO-301
514 02 20 0001

TA-TO-303
740 05 101

TEL.HO500
505 05 10 0023

TEL.HO1000
505 05 10 0024

TEL.HO1500
505 05 10 0025

TEL.HO2000
505 05 10 0026

Lapping plate•Set of abrasive oil for probe lapping••• • • • • • •Protection cover for instruments
500 mm

•

• •

• •

Protection cover for instruments
1000 mm

•

• •

• •

Protection cover for instruments
1500 mm

•

• •

• •

Protection cover for instruments
2000 mm•• ••Protection cover for instruments
3000 mm• • •Protection cover for instruments
4000 mm•Protection cover for instruments
5000 mm•Protection cover for instruments
6000 mm•

505 05 10 0032

Protection cover for instruments
8000 mm•

TEL70.3

Display unit for TEL70 + TULM70• •TEL.HO3000
505 05 10 0027

TEL.HO4000
505 05 10 0028

TEL.HO5000
505 05 10 0029

TEL.HO6000
505 05 10 0030

TEL.HO8000

351 191 0001

109

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

EL-D80S

351 101 1001

Display unit SYLVAC D100S••

351 201 0012

Display unit Quadra-Chek ND 1100
(2 axes)•

TA-DU-330

Display unit Heidenhain ND287

TA-DU-303

Foot pedal

3706 0002

TA-EL-301
3706 0009

Foot pedal for Quadra-chek QC110/QC120

TA-EL-040

Joystick•• • • •
• •••358 0020

TVM.O-PC/
AT.9P

LABCN

• •

351 192 0045

TELMA31LABC + P•

Display unit Quadra-Chek ND 1100,(1 Vpp) with standard foot• •351 101 1011

THV•

Display unit Quadra-Chek ND 1100,(1 Vpp) with OEM footTA-DU-302

HPD

HPA

HG

•

351 292 0010

TA-DU-301

H

Display unit SYLVAC D80S

351 292 0011

EL-D100S

ALESTATELSACCESSOries

Cable Opto-PC/AT 9 P/F 2 m••

USB-RS232 Converter•• •Laser printer (USB)•• •Inkjet printer (USB)•• •333 9 0003

V-31

Cable for RS232 printer•Cable RS232 m/f, 1.8 m,
VT/M/MT: Printer + V+/H : PC•

333 0 0003

CABL.RS.1/
1-9P
332 01 0001

TA-EL-011
358 0006

LABC-40356 0010

TA-EL-030
356 0016

110

LABCNLABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries

Printer (RS232)•

Paper rolls (5 pces)•

Power cable, 2 poles, Europe

•

•

•

Power cable, 2 poles, USA/Japan

•

•

•

Power cable, 2 poles, Australia

•

•

•

Power cable, 2 poles, UK

•

•

•

Power cable, 2 poles, Korea

•

•

•

Universal AC adapter, 9 V(without cable), Suitable for V+, H•

Battery• ••Ext. power supply for force display
HP/LABC• ••Temperature compensation system,with 2 temperature sensors•• •Temperature compensation system,with 3 temperature sensors•• •Temperature compensation system for
TEMPCOMP-PA laboratory, with WinComp Advanced
609 50 003
and 9 sensors, (7x temperature, 1x
humidity, 1x pressure)•• •Temperature compensation system forlaboratory, with WinComp Advanced
and 9 sensors, (7x temperature, 1x
humidity, 1x pressure)with DKD certificate

•

• •

V-30
356 0007

V-30.7788 000001 001

TA-EL-001
332 10 0011

TA-EL-002
332 10 0013

TA-EL-003
332 10 0016

TA-EL-004
332 10 0014

TA-EL-005
616 20 003

TA-EL-131
334 0020

BAT-TVM.
OPTO
3705 0002

TA-EL-331
740 16 002

TEMPCOMP-B
609 50 001

TEMPCOMP-P
609 50 002

TEMPCOMPPA DKD
609 50 004

111

TRIMOS | HORIZONTAL MEASURING INSTRUMENTS

TELS

Alesta

H

HG

HPA

HPD

THV

LABC + P

LABCN

TA-SW-001

Software for data transfer WinDDE

394 1 0050

• • •

LABCN

LABC + P

THV

HPD

HPA

HG

H

ALESTATELSACCESSOries•

Gauge management software• • • •

Module for inspection of calipers•

Module for inspection of dial gauges• • • •

Module for inspection of micrometers•

Module for inspection of pins• • • •

394 1 3206

Module for inspection of plain gauges ,Plug gauges, ring gauges, snap gauges• • • •QM-BLOCK

Module for inspection of gauge blocks• • • •

Module for inspection of threadgauges• • • •

Module for inspection of taper threadgauges•• •Module for inspection of spline
gauges•• •QM-MANAG
394 1 3201

QM-CALIP
394 1 3202

QM-DIAL
394 1 3203

QM-MICRO394 1 3204

QM-PIN394 1 3205

QM-PLAIN

394 1 3210

QM-THREAD
394 1 3212

QM-TTAP32
394 1 3214

QM-SPLINE
394 1 3213

QM-THREAD32 Starter Module for inspection of
STARTER
thread gauges

• •

• •

• • • •

394 1 3222

QM-TTAP32
STARTER

Starter Module for inspection of taper
thread gauges•• •Starter package including, QM-MANAG Light, QM-THREAD, QM-PLAIN ,no extension available• • • •

Standard package including:
QM-MANAG, QM-PLAIN,
QM-THREAD, QM-DIAL,QMCALIP, QM-MICRO• • • •

394 1 3224

QM-PACK32-1
394 1 3250

QM-PACK32-2
394 1 3251

112