DIGITAL BLOOD PRESSURE MONITOR

1901

Contents................
Blood Pressure Monitor Intended Use
02....... 1. What is blood pressure ?
02....... 2. Why is it useful to measure blood
01.......

..........
..........
05.......

pressure at home ?
A. WHO blood pressure classifications
B. Variations in blood pressure

3. Important information before use of thedevice08....... 4. Device description..........
A. Name of the parts..........
B. Description of display symbols

5. Changing Batteries
11....... 6. Attaching pressure cuff
12....... 7. Correct measurement position
10.......

A

Contents................
8. Symbol Difinitions.
15....... 9. How to measure.
13.......

..........
..........
..........
..........
..........

A. Inserting batteries in position.
B. Setting date and time.
C. Steps to take blood pressure
measurement.
D. Storing recalling and erasing
measurement data.
E. To change batteries.

10. Maintenance.
19....... 11. Safe-keeping.
19....... 12. To prevent malfunction.
20....... 13. Warning.
21....... 14. Specifications.
18.......

B

MEDICAL DISCLAIMER

This manual and product are not meant to be a
substitute for advice provided by your doctor or othermedical professional.
You are not to use the information contained herein,or this product for diagnosing or treating a healthproblem or prescribing any medication. If you have or
suspect that you have a medical problem, promptly
consult your healthcare provider.

Blood Pressure Monitor Intended Use:
Measures human's Systolic blood pressure, Diastolic bloodpressure and pulse on wrist by oscillometric measurement
method.
The device is designed for home use and recommended foruse by adults aged 18 years and older with wrist circumference
ranging 135 ~ 195 mm (5.3~7.7 inch).

1

1. What is blood pressure ?

Blood pressure is a measurement of the force of blood flowing
against the walls of the arteries. Arterial blood pressure isconstantly changing during the course of the cardiac cycle. The
highest pressure in the cycle is called the systolic blood pressure,the lowest is the diastolic blood pressure. Both pressure readings,
the systolic and the diastolic are necessary to enable a physician
to evaluate the status of a patient's blood pressure. Many factors
such as physical activity, anxiety or the time of day, can influence
your blood pressure. Blood pressure is typically low in themorning and increases from the afternoon to the evening. It is on
average lower in the summer and higher in the winter.
Medical practitioners generally recommend the “Rule of 3”, where
you are encouraged to take your blood pressure three times in arow (at 1 minute interval), three times a day for three days. After
three days you can average all the results and this will give youan accurate idea of what your blood pressure really is.

2. Why is it useful to measure blood pressure
at home ?

Having one's blood pressure measured by a doctor in a
hospital or a clinic, and a group health checks, tend to stimulate
nervousness in the subject and may even create high blood2

pressure. Also varies blood pressure in accordance with a varietyof conditions and so judgment is not possible on the basis of asingle measurement.
The blood pressure measured first thing in the morning aftergetting up, before taking any food and with the subject still,is known as the fundamental blood pressure. In practice it israther difficult to record the fundamental blood pressure, but tocome as near as possible to measuring the blood pressure in
an environment that is close to this, is why it is useful to take themeasurement at home.
A. WHO blood pressure classifications
Standards for assessmentof high or low bloodpressure without regard toage, have been establishedby the World HealthOrganization (WHO), as
shown in the chart.

* Figure No.1

This chart is intended to be used as a guide in understanding
noninvasive blood pressure measurements. Please consult withyour physician for proper diagnosis.

3B. Variations in blood pressure
Individual blood pressures vary greatly both on a daily and aseasonal basis. These variations are even more pronounced in
hyper tense patients. Normally the blood pressure rises while atwork and is at its lowest during sleeping period.
(Hyper tense: means a person who has high blood pressure
symptom.)
The graph below illustrated the variations in blood pressure overa whole day with measurement taken every five minutes.The thickline represents sleep. The rise in blood pressure at 4 PM (A inthe graph) and 12 PM (B in the graph) correspond to an attack of
pain.

(Direct arterial pressure recording in unrestricted
man. Beven, Honour & Stott: Clin. Sci. 36:329.
1969)* Figure No.2

4

3. Important information before use of the device* Do not use this manual and product as a substitute foradvice, diagnosing or treating a health problem or prescribing
any medication by your doctor.
* The device is not suitable for clinical use.
* This monitor is not intended for use in the MR environment.
Please rest for at least 5~10 minutes before taking themeasurement.
To allow your blood vessels to return to the condition prior to
taking the measurement, please wait at least 3~5 minutes
between measurements. You may need to adjust the wait time
according to your personal physiological situation.
If you have one of the circulatory problems as arteriosclerosis,
diabetes, liver disease, kidney disease, severe hypertension,
peripheral circulation and medical problem, please consult
your doctor or healthcare professional before using the device.
Wait 30 ~ 45 minutes before measurement if you’ve justconsumed caffeinated beverages or smoked cigarettes.
Blood pressure measurements should be interpreted by aphysician or trained health professional who is familiar withyour medical history. By using the unit regularly and recording
the results for your physician to interpret, you keep yourphysician informed of the continuing trends in your bloodpressure.
Wind the cuff snugly around your wrist and the cuff must be at
the same level as your heart.
5

Do not vibrate the unit during measurement, or the proper
measurement will not be achieved.
Perform measurement quietly in a relaxed position.
Do not wind the cuff over jacket or sweater sleeve, ormeasurement cannot be done.
Keep in mind, that blood pressure naturally varies from time totime throughout the day and also is affected by lots of different
factors such as smoking, alcohol consumption, medication andphysical activity.
People with a condition that causes circulatory problems(diabetes, kidney disease, arteriosclerosis or poor peripheral
circulation) may get lower readings with this monitor that with ablood pressure monitor that is used on the upper arm. Please
consult your physician to determine if your wrist blood pressure
accurately reflects your actual blood pressure.
Blood pressure measurement determine with the unit areequivalent to those obtained by a trained observer using
the cuff / stethoscope auscultation method. Within the limitsprescribed by the Standard of EN 1060-4.
We recommend you using the same wrist (preferably the left
wrist) and measuring around the same time each day.
Perform measurements in a quiet and relaxed environment atroom temperature.
This product is not suitable for:Pregnant women
People with arrhythmias
Undergoing intravenous injection on any limb
Currently in a dialysis treatment
In pre-eclampsia condition
6

For those who have had mastectomy surgery (especiallywhose’ lymph nodes removed), it’s recommend take ameasurement on the unaffected side.
When used among medical electronic equipments on thesame limb, pressurization of the cuff may cause temporarily
malfunction to other devices.
*Note!
1. Do not use the device on infants, children, or those who
cannot express their own intention.
2. The device is equipped with sensitive electronic components.
While measuring, avoid strong electrical or electromagnetic
fields, e.g. mobile phones, microwave ovens, etc; or it may
lead to temporary reading error or inaccuracy.
3. Consider the electromagnetic compatibility of the device (ex.
power disturbance, radio frequency interference etc.) Please
use it indoor only.
4. Over high frequency measurements may result in bloodflow interference, which is likely to cause uncomfortablesensations, such as partial subcutaneous hemorrhage,or temporary numbness to your wrist. In general, these
symptoms should not last long. However, if you do notrecover in time, please seek your medical practitioners for
help.

7

4. Device description:
A. Name of the parts

LCD DISPLAYSETTING BUTTON

AVG

SY

mm

FUNCTION
BUTTONM RECALLMEMORY BUTTON

S

Hg

:

DI

PM

M

A
H

mm

g

Pu

ST
AR
T
ST
OP

lse

/m

in

START/STOP
BUTTON(
Stand-by)
WRIST CUFF

*Caution!
Any artificial changes to theoriginal device might result inmeasurement error.
Substitution of a component
different from that supplied
might result in measurement
error.

CARRYINGCASE

ALKALINE BATTERY

UM-4 (AAA) LR03 1.5 V x 2

8* Figure No.3

B. Description of display symbols

SYSTOLICBLOOD
PRESSURE
LAST 3 READINGAVERAGEAVG

DIASTOLIC
BLOOD
PRESSURE

LOW BATTERYSYMBOLHOUR：MINUTE(ALSO REPRESENTS AS YEARUNDER SETTING MODE):

PM

MONTH/DATEPULSE SYMBOL

9

PULSE

5. Changing batteries

Remove the battery cover at the left side of the unit and insert
batteries into the battery compartment in accordance with ribbon
strip as shown, taking extreme care that the polarities + and - areobserved. Important notice: to take the batteries out easily, justpull the ribbon strip for help.

* Figure No.4

Remove the battery cover and insert 2 AAA LR03 alkaline
batteries into the battery compartment as shown.
*Note!
● Batteries are hazardous waste. Do not dispose of them
together with the household garbage. Please take the usedbatteries to the recycling collection point according to yourlocal regulations.
● Keep the battery away from small children in case theyswallow it.
● To prolong the battery life and prevent damage caused byleakage, remove the batteries from the device if the device
is not to be used for a long period.
● Memories (if any) will not be deleted during battery
replacement.
● After replacing the batteries, reset the date and time.
10

6. Attaching pressure cuff

A. Wrap the pressure cuff around the wrist.
a. The display of the unit should beplaced on the palm side of the wrist.
b. The wrist should be bare.
B. Fasten the pressure cuff snugly.

10 mm(0.4 inch)* Figure No.5

a. Do not pull strongly on the pressure
cuff.
b. Do not make the pressure cuff tootight.
c. Leave approximately 10 mm (0.4inches) between the sleeve and thebottom of the palm of your hand.
Figure No. 5.

135 ~ 195 mm(5.3 ~ 7.7 inch)* Figure No.5-1*Note!
● Do not use this device if your wrist has any wound or injury.
● Do not wrap the cuff around any body part other than yourwrist.

11

7. Correct measurement position

It is extremely important for the measured values, that the cuff isat the same height of the heart.
Higher or deeper position will give deviations from the true value.
Measurement while sitting down:a. Sit down comfortably with your feet flaton the floor.
Please do not cross your legs during
measurements.
b. Position the blood pressure monitor on
your wrist.
c. Place your elbow on a table or otherobject.
d. Use the armrest to position the arm sothat the pressure cuff is at the sameheight as the heart.
e. Relax your hand with the palm facing up.

12* Figure No.6

8. Symbol Definitions
SYMBOLS
AVG

AVG

Low Battery
SymbolDefinitions
This symbol appears when the battery poweris excessively low or the polarity reverses.
➔ We suggest you replace all batteries withnew ones, and make sure the +/- polarities areproperly positioned.
Once pulse is detected, the symbol flashes
with each pulse beat.

➔ Our suggestion:
Pulse Symbol Please do not talk or move during
measurements.

AVG
Last 3 Reading
Average

This symbol appears when LCD displays
average value of last 3 readings.

13

SYMBOLS/SYMPTOMS

CONDITIONS/CAUSES

INDICATION/
CORRECTION

Worn-out batteries.

Replace them with
2 new AAA (LR03)
alkaline batteries.

Battery polarities
have been positioned
incorrectly.

Re-insert the batteries
in the correct
positions.

AVG Measuring Error Cuff has been placed
incorrectly.

Wrap the cuff properly
so that it is positioned
correctly.

Did you talk or move
during measurement?

Measure again. Keepwrist steady during
measurement.

No power-on by

AVGpressing

buttonSymbol shown
AVG on display

or the bloodpressure value
is displayedexcessively low
or high.
Measuring
Error Symbol
Measuring
Error Symbol
Measuring
Error Symbol

Shaking of the wristwith the cuff on.
Air circuit adnormality.
Cuff tube may not be
plugged into monitor
correctly.
Inflation pressure
exceeding 300 mmHg.

Check cuff connection.
Measure again.

Can't determine bloodpressure measurement
data.

Wrap the cuff properly
and keep steady.
Measure again.

Switch the unit off,
then measure again.

Note: If "EP" appears on the display, just return the device to your local
distributor or importer.
14

9. How to measure

A. Inserting batteries in position:a. Open battery cover.
b. Refer to the picture in positioning batteries. (Figure No.4)
c. All LCD segments appear on the display in 3 seconds.
B. Setting Date and Time:a. Press "" button over 2 secs ("year" starts flashing)Press "" button to set the correct year (2011,2012,…2026)
b. Press "" button ("month" starts flashing)Press "" button to set the correct month (1,2,3,.....12)c. Press "" button again ("date" starts flashing)Press "" button to set the correct dated. Press "" button again ("hour" starts flashing)Press "" button to set the correct hour in 12-hour format
e. Press "" button again ("minute" starts flashing)Press "" button to set the correct minute (0,1,2,3,.....59)
f. Press "" button again to switch to normal time(month, date, hour and minute appear on the display)
C. Steps to take blood pressure measurement:a. Wrap the cuff around the wrist (Refer to Wrapping of Wrist
cuff as Figure No. 5)
b. Sit upright on the chair to have correct posture. (Refer toCorrect Posture in Taking Blood Pressure as (Figure No. 6)"button. All elements display '888' in 3 seconds.
c. Press"
Afterward, displays the updated time. Pulse symbol will flash
15

on the display and then the air pressure will automaticallypump up to 195 mmHg. If blood pressure tends to be on thehigh level, the air pressure will automatically increase to therequired level for measurement.
*Note!
Do not inflate the cuff until it is wrapped around your wrist.
* Do not move or talk in the midst of taking blood pressure
measurement.
d. After the air pressure is increased, it will slowly decrease.
When the pulse is detected,
will start flashing.
*Note!
• If the cuff does not stop inflating, remove the cuff at once.
• To interrupt the measurement, you may press the
button.
The cuff will deflate immediately after a button is pressed.
e. After taking blood pressure measurement, the Systolic bloodpressure, Diastolic blood pressure and pulse will be on thedisplay for 1 minute.
D. Storing, recalling and erasing measurement data:a. Storing data:
After each blood pressure measurement, the Systolic bloodpressure, diastolic blood pressure, pulse and the time & dateof specific day will be automatically stored. The memory
holds the latest 99 measurement data. If more than 99
measurements, the memory will automatically clear out theearliest data.
16

b. Recalling data:(1). Press " M " button. If there is no data in the memory, only
TIME /DATE will appear on the display. If yes, the firstreading will be the average of the latest 3 measurements.
(The average value dose not appear if there are two or fewer
measurements stored in memory.)(2). Press " M " button again, the latest measurement willappear with the number as well as the recording date
and time. The data called up on the display is numberedaccordingly. After reading the recalled data as rotator
99-set of the stored data, press " M " to switch to normal
time.
(3). To discontinue in the midst of reading of the data, press"" button.
c. Erasing data:(1). Press the "M" button.
(2). Press hold "" and "" button to clear memories.
"CLA" show up on the display for 3 seconds.
Note: Once deleted, your data can NOT be restored.
E. To change batteries:" show up on the display, the battery is weak.
When "
It will not be possible to take blood pressure measurement.
All batteries used must be the same type. Do not mix alkaline,
standard(carbon-zinc) or rechargeable (cadmium) batteries.
Do not mix old and new batteries.
Please replace both batteries together.
17

10.MAINTENANCE:

Use a piece of cloth with water or mild cleansing agent to wipe
the device and dry it immediately with a dry cloth.
Do not use detergent or any strong chemicals to clean thedevice.
Disinfection - Use a piece of cloth with a dilution of neutral
detergent or soapsuds to wipe the surface of the cuff for 10
seconds.
Make sure the cuff is completely dry before using.
If any suggestion or service is requested, please consult yourlocal distributor.
Do not implement the maintenance procedures for device
during measurement.
Only trained technicians are allowed to repair and dissemble
the device, including software upgrades, patches andmaintenance.

*Note!
Water quality required for cleaning and disinfection: Tap water.

18

11.SAFE-KEEPING:

Always keep the device in the carrying case after its use.
Do not put the item directly under the sunlight, in hightemperature, or humid and dusty places.
Do not store the device in extremely low (less than –13 °F/–25 °C) and high (more than 158 °F/ 70 °C) temperature, nor
in a place its humidity exceeds 93% R.H.

12.TO PREVENT MALFUNCTION:

Do not in any way twist the wrist cuff.
Do not start the air pressure when the wrist cuff is not wrappedaround the wrist.
Do not attempt to disassemble or change any parts of themonitor including wrist cuff.
Do not drop the product or put it through strong impact.

19

13.WARNING:

The device measures the blood pressure of adult only.
In the event of the cuff pumping up and not stopping, please
open the cuff at once.
To ensure precise measurement reading, recalibration of
the device is recommended after 2 years from the date ofpurchase. Shipping plus handling cost and recalibration
service fee shall be charged accordingly.

Limited WarrantyThis device is warranted to be free from manufacturing
defects for a period of 1 year from the date of purchase.
Within the warranty period, should there be any manufacturing
defect found and repair is needed, please contact your local
distributor accordingly.
Please note warranty does not cover damage caused by,including but not limited to: misuse or abuse; accident; theattachment of any unauthorized accessory; alteration to
the product; improper installation; unauthorized repair ormodification; improper use of electrical/ power supply; dropped
product; failure to follow required maintenance and safekeeping; transportation damage.

20

14.SPECIFICATIONS:

Model Number:CH-617
Method of Measurement :OscillometricRange of Measurement :Pressure 0 ~ 300 mmHg
Pulse 40 ~ 199 beats/minuteAccuracy:Pressure +/- 3 mmHg,
Pulse +/- 5% of reading
Inflation:Automatic inflation (Air Pump)
Deflation of Pressure:Automatic air release control valve
Display:Liquid Crystal Digital Display
Sets of Memory:99 sets
Unit Dimension:L70XW72XH31mm
L2.75 x W2.83 x H1.22 inch
Unit Weight:88 ± 5 g (3.10 ± 0.17 oz) (Batteries and cuff excluded)
Cuff Size:Wrist circumference approx. 13.5 - 19.5 cm
Storage/Transportation :Temperature: -25 °C ~ 70 °C (-13 °F ~ 158 °F)
Environment
Humidity: ≤ 93 % R.H.
Operation Environment :Temperature: 5 °C ~ 40 °C (41 °F ~ 104 °F)
Humidity: 15 % ~ 93 % R.H.
Atmospheric pressure: 700 hPa ~ 1060 hPa
Auto power off:Whenever not used for 1 minute
Product Life
:5 Years (4 times per day)Power Supply:2 x AAA LR03 (2 x 1.5 V) Alkaline battery (3V DC)
Battery Life
:Approx. 250 times measurement
Accessories:Carrying case, instruction manual, 2 pcs AAA LR03 battery

These specifications are subject to change without notice for purpose of improvement.
21

mentsests

ements

s to
non-

Note:

This blood pressure monitor complies with the EC

0197 Directive (93/42/EEC) and bears the CE mark. This
blood pressure monitor also complies with mainly
following standards,(included but not limited)

Safety standard:
EN60601-1
Medical electrical equipment Part 1: General requirements forbasic safety and essential performance
EMC standard:
EN 60601-1-2 Medical electrical equipment Part 1-2: General requirements forbasic safety and essential performance - Collateral Standard:
Electromagnetic disturbances - Requirements and tests
Performance standards:
EN 1060-1
Non-invasive sphygmomanometers - Part 1: General
requirements
EN 1060-3
Non-invasive sphygmomanometers - Part 3: Supplementary
requirements for electro-mechanical blood pressure measuringsystems
EN 1060-4
Non-invasive sphygmomanometers - Part 4: Test procedures
to determine the overall system accuracy of automated noninvasive sphygmomanometers
EN ISO 81060-1 Non-invasive sphygmomanometers - Part 1: Requirements andtest methods for non-automated measurement type (partially applied)
IEC 80601-2-30 Medical electrical equipment - Part 2-30: Particularrequirements for the basic safety and essential performance of automated noninvasive sphygmomanometers
Follow instructions for use.
22

Date of Manufacture
TYPE BF APPLIED PART- Internally powered equipment- Not suitable for use in presence of flammable anaesthetic
mixture with air or with Oxygen or nitrous oxide- Continuous operation with short-time loading
Degrees of protection provided by enclosures
To avoid inaccurate results caused by electromagnetic
interference between electrical and electronic equipments, do
not use the device near a cell phone or microwave oven.
At least keep a maximum output power of 2 W yields and a
distance 3.3m away from this equipment.
Discard the used product to the recycling collection pointaccording to local regulations.
Manufacturer:
HEALTH & LIFE Co., Ltd.
9F, No. 186, Jian Yi Road, Zhonghe District, New Taipei City,
Taiwanwww.healthandlife.com.tw
Authorized Representative in the European Community:
EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
23Serial Number

LOT

Batch code
Atmospheric pressure limitation
Humidity limitation
Temperature limitation

WEEE MARK
If you want to dispose this product, do not mix with generalhousehold waste. There is a separate collection systems forused electronics products in accordance with legislation under
the WEEE Directive (Directive 2012/19/EU) and is effective only
within European Union.
Information for Users on Collection and Disposal of used Batteries.
The symbol in this information sheet means that used batteries
should not be mixed with general household waste.
For proper treatment,
recovery and recycling of used batteries,
0197
please take them to applicable collection points.
Pb
For more information about collection and recycling of batteries,
please contact your local municipality, your waste disposal service
or the point of sale where you purchased the items.
24

0537

CONTENIDO.......................
Uso del monitor de tensión arterial
02....... 1. ¿Qué es la presión arterial?
02....... 2. ¿Por qué es útil medir la presión arterial
01.......

..........
..........
05.......

en la casa?
A. C
 lasificaciones de presión arterial
según la organización mundial de lasalud (OMS)
B. Variaciones de la presión arterial

3. Información importante antes de utilizar
el equipo

08....... 4. Descripción del equipo.
..........
A. Listado de componentes..........
B. D
 escripción de Simbolos dePresentación.

5. Cambio de pilas.
11....... 6. Colocación del brazalete
12....... 7. Posición de medición correcta.
10.......

A

CONTENIDO.......................
8. Definiciones de símbolos
15....... 9. ¿Cómo medir?
13.......

..........
..........
..........
..........
..........

A. Insertar pilas en posición.
B. Puesta de Fecha y Horas.
C. P
 asos para tomar la medición depresión arterial.
D. G rabación, ubicación y borradura dedatos de medida.
E. Cambio de pilas.

10. Mantenimiento.
19....... 11. Medidas de seguridad.
19....... 12. Para prevenir el mal funcionamiento
20....... 13. Advertencia.
21....... 14. Especificaciones.
18.......

B

RENUNCIA DERESPONSABILIDAD MÉDICA
Este manual y este producto no pretenden ser
un sustituto del consejo de su médico ni de otro
profesional de la salud.
No utilice la información aquí contenida, ni este
producto para diagnosticar ni tratar problemas desalud, ni para recetar medicamentos.
Si tiene o sospecha que tiene un problema de salud,consúltelo cuanto antes con su médico.

Uso del monitor de tensión arterial:
Mide la tensión arterial sistólica, la tensión arterial diastólica y elpulso en los humanos a través de la muñeca mediante un método
de medición oscilométrico.
El dispositivo está diseñado para uso doméstico y recomendado
para que lo utilicen adultos mayores de 18 años y una
circunferencia de muñeca comprendida entre 135 y 195 mm(5,3~7,7 pulgadas).

11. ¿Qué es la presión arterial?

La presión arterial es una medida de la fuerza que ejerce lasangre sobre las paredes de las arterias cuando circula por elcuerpo. La presión de sangre arterial cambia en forma constante
durante el curso del ciclo cardíaco. La presión más alta en el
ciclo es conocida como presión sistólica mientras la más baja
es la diastólica. Las dos lecturas de sangre, la sistólica y ladiastólica son necesarias para que un medico evalue el estado
de la presión de sangre de su paciente. Muchos factores, tales
come actividad física, ansiedad o tiempo del día, pueden influir su
presión de sangre. La presión desangre está baja en la mañana
y empieza a aumentar desde la tarde hasta la noche. Es enpromedio más bajo en verano y más alto en invierno.
Los médicos generalmente recomiendan la “Regla del 3”, donde
se le anima a medir la presión arterial tres veces seguidas (aintervalos de 1 minuto), tres veces al día durante tres días.
Después de tres días puede promediar todos los resultados, lo
que le dará una idea precisa de cuál es realmente su presión
arterial.

2. ¿Por qué es útil medir la presión arterial enla casa?

La medición de la presión arterial por parte de un médico en un
hospital o una clínica, y los controles de salud grupales, tienden aestimular el nerviosismo en el sujeto e incluso pueden producir una
presión arterial alta.
2

La presión arterial también varía en función de una variedad
de condiciones y por lo tanto no es posible llevar a cabo una
valoración sobre la base de una sola medición.
La presión arterial medida a primera hora de la mañana despuésde levantarse, antes de tomar cualquier alimento y con la persona
quieta, se conoce como la presión arterial fundamental.
En la práctica, resulta más dificil registrar la presión arterial
fundamental, pero acercarse lo más posible a la medición unentorno cercano explica que es útil tomar la medida en casa.
A. Clasificaciones de presión arterial según la organización
mundial de la salud (OMS)
Los standards usados parala evaluación de presión
alta y baja sin importar laedad, han sido establecido
por la Organización Mundialde Salud (OMS) y se dan acontinuación:* Figura No.1

Esta tabla está diseñada para usarse como guía para entenderlas mediciones no invasivas de la presión arterial. Consulte a su
médico para un diagnóstico adecuado.

3

B. Variaciones de la presión arterial.
"La presión arterial varía durante el transcurso del día y el año.
Estas variaciones son aún más pronunciada en pacientes que
sufren de hipertensión*."Normalmente la presión arterial aumenta durante el trabajo y seencuentra en el nivel más bajo durante el período de sueño.
(Hipertensión: lo que indica que una persona que tiene altapresión arterial)
La grafica dada a continuación indica la variación de la presiónde sangre en todo el día con medidas tomadas cada 5 minutos.
La línea gruesa indica sueño. La subida en la presión de sangre
a las 4:00 PM (A en la gráfica) y 12:00 PM (B en la gráfica)corresponde a un ataque.

(Registro directo de la presión arterial en elhombre sin restricciones. Beven, Honour y Stott:Clin. Sci. 36:329. 1969)* Figura No.2

4

3. Información importante antes de utilizar el equipo* No utilice este manual ni el producto como un sustituto para
aviso, diagnóstico o tratamiento de un problema de salud o pararecetar cualquier medicamento que deba prescribir un médico.
* El dispositivo no es adecuado para uso clínico.
* Este monitor no está diseñado para utilizarse en el entorno MR.
Por favor, descanse al menos 5~10 minutos antes de realizar la
medición.
Para que sus vasos sanguíneos retornen a su estado normal antes
de realizar una medición, espere al menos 3~5 minutos entre una
medición y la siguiente. Es posible que necesite ajustar el tiempode espera de acuerdo con su situación fisiológica personal.
Si tiene algún problema circulatorio como arterioesclerosis,
diabetes, enfermedad hepática, enfermedad renal, hipertensión
severa, circulación periférica y un problema médico, consultea su médico o a un profesional de la salud antes de usar este
dispositivo.
Espere entre 30 ~ 45 minutos antes de medir la presión luego dehaber consumido bebidas a base de cafeína o si ha fumado.
Las mediciones de la presión arterial deben ser interpretadas
por un médico o profesional de la salud capacitado que estéfamiliarizado con su historial médico.
Al usar el equipo de forma regular y registrar los resultados paraque su médico los interprete, usted lo mantiene informado de lasvariaciones en su presión arterial.
Enrolle la manga cómodamente alrededor de su muñeca. La
muñeca debe estar al mismo nivel que su corazón.
5No mueva el equipo durante proceso, o la medición no se logrará.
Lleve a cabo la medición en silencio en una posición relajada.
No coloque la manga por sobre la ropa, porque la medición nopodrá realizarse.
Acúerdese de que la presión de sangre varía en forma naturaly regularmente durante el día y es afectada por varios factorestales como el fumar cigarrillos, tomar alcohol, medicamentos y
actividades médicas.
La gente con una condición que cause problemas en circulación(diabetes, enfermedad en los riñones, arterioclorósis o baja
circulación periféral) hará que las lecturas sean más bajas con este
tensiómetro que con un tensiómetro de brazo. Consulte con su
médico para determinar si la presión arterial medida en su muñeca
es confiable.
La medición de la presión arterial determinada con la unidad
equivale a los obtenidos por un observador entrenado usando
el método de auscultación de puño / estetoscopio. Dentro de loslímites establecidos en la directiva EN 1060-4.
Le recomendamos utilizar siempre la misma muñeca(preferiblemente la izquierda) y realizar la medición
aproximadamente a la misma hora del día.
Realice las mediciones en un entorno tranquilo y relajado atemperatura ambiente.
Este producto no es adecuado para:
Mujeres embarazadas
Personas con arritmias
Realizar una inyección intravenosa en cualquier extremidad
Actualmente en tratamiento de diálisis
En estado de preeclapsia
6Para quienes han sufrido una cirugía de mastectomía(especialmente a quienes se les ha quitado nodos linfáticos), es
recomendable tomar una medición en el lado no afectado.
Si se utiliza en conjunto con otros dispositivos médicos en lamisma extremidad, la presurización de la manga puede causar un
mal funcionamiento temporal en otros dispositivos.
*¡Nota!
1. No utilice el dispositivo en los bebés, niños, o aquellos que nopueden expresar su propia intención.
2. El dispositivo está equipado con componentes electrónicos
sensibles.
Durante la medición, evite campos eléctricos oelectromagnéticos intensos, por ejemplo, teléfonos móviles,
hornos de microondas, etc.; si no sigue esta recomendación, sepueden producir errores de lectura o imprecisiones temporales.
3. Tenga en cuenta la compatibilidad electromagnética deldispositivo (por ejemplo, perturbaciones de la alimentación,
interferencia de radiofrecuencia, etc.). Utilícelo solo eninteriores.
4. Durante las mediciones de alta frecuencia puede provocar
interferencias en el flujo sanguíneo, lo cual es probabilidadesde causar sensaciones desagradables, tales como parcial por
vía subcutánea hemorragia, o entumecimiento temporal a su
muñeca. En general, estos síntomas no deben durar mucho
tiempo. Sin embargo, si no se recuperan en el tiempo, por favor
busque ayuda médica.
7

4. Descripción del equipo:
A. Listado de componentes

Display LCD
BOTÓN

AJUSTE

AVG

SY

mm

S

Hg

:

Tecla Función

M Tecla memoria*¡Precaución!
Cualquier cambio
artificial en el
dispositivo original
podría provocar unerror en la medición.
La sustitución de uncomponente diferente
al suministrado podría
provocar un error de
medición.

DI

PM

M

A
H

mm

g

Pu

ST
AR
T
ST
OP

lse

/m

in

Tecla Empezar/
Parar
Manga

EstuchePILA ALCALINA
UM-4 (AAA) LR03 1,5 V x 2

8* Figura No.3

B. Descripción de Simbolos de Presentación

TENSIÓN
ARTERIALSISTÓLICAPROMEDIO DE LASÚLTIMAS 3 MEDICIONESAVGTENSIÓN
ARTERIAL
DIASTÓLICA

SÍMBOLO DE CARGABAJA DE LAS PILASHORA: MINUTO(TAMBIÉN REPRESENTA EL AÑO EN
EL MODO DE CONFIGURACIÓN):PM

MES/DÍA
SÍMBOLO DE PULSO

9

PULSO

5. Cambio de pilas

Retire la tapa de las pilas en el lado izquierdo de la unidad e
introduzca las pilas en el compartimento de pilas de acuerdo
con la cinta, tal y como se indica, teniendo cuidado de que laspolaridades + y - se respeten.
Notas importantes: ud.
podrá tener dificultades
en retirar las pilas.
En este caso, usa unartículo duro y pequeño
como un lápiz u otros
como ayuda.

* Figura No.4

Retire la tapa de las pilas e inserte 2 pilas alcalinas LR03 AAA enel compartimento de las pilas tal y como se muestra.
*¡Nota!
● Las pilas son residuos peligrosos. No se deshaga de ellasarrojándolas a la basura doméstica. Lleve las pilas usadas aun punto de reciclaje cumpliendo las normativas locales.
● Mantenga las pilas alejadas de los niños pequeños paraevitar que se las traguen.
● Para prolongar la vida útil de las pilas y evitar daños
causados por fuga del líquido de las mismas, retire las pilasdel equipo si no va a utilizar este durante un período detiempo prolongado.
● Las memorias (si las hay) no se borrarán durante lasustitución de las pilas.
● Al sustituir las pilas, se restablecerá la fecha y hora.
106. Colocación del brazalete

A. Enrolle la manga del tensiómetro enla muñeca.
a. El display del equipo debe
colocarse en el lado de la palmade mano.

10 mm(0,4 pulgadas)b. La muñeca debe estar sin ropa.
B. Coloque la manga firmemente.

* Figura No.5

a. No tire mucho de la manga.
b. No presione mucho la manga.
c. Deje aproximadamente 10 mm(0,4 pulgadas aproximadamente)
entre la manga y la parte inferior dela palma de la mano. Figura No. 5

135 ~ 195 mm(5,3 ~ 7,7 pulgadas)* Figura No.5-1*¡Nota!
● No utilice este dispositivo si la muñeca tiene alguna herida olesión.
● No envuelva la manga alrededor de ninguna parte delcuerpo que no sea la muñeca.
11

7. Posición de medición correcta

Es extremadamente importante para los valores medidos, que lamuñeca esté a la misma altura del corazon.
Una posición más alta o baja dará desviaciones del valor
verdadero.
Medición sentado:a. Siéntese cómodamente con los pies
apoyados sobre el suelo.
No cruce las piernas durante las
mediciones.
b. Coloque el tensiómetro en la muñeca.
c. Coloque su codo sobre una mesa.
d. Utilice el reposabrazos para colocar
el brazo de tal modo que el brazalete
se encuentre a la misma altura que elcorazón.
e. Relaje su mano con la palma hacia
arriba -- llustracion -- (Figura No.6)

12* Figura No.6

8. Definiciones de símbolos
AVGSÍMBOLOS
AVG

Símbolo decarga bajade las pilas

Definiciones
Este símbolo aparece cuando la carga de laspilas es excesivamente baja o la polaridad está
invertida.
➔ Es recomendable cambiar todas las pilas
por otras nuevas y asegurarse de que laspolaridades +/- están colocadas correctamente.
Cuando se detecte el pulso, el símbolo
parpadeará con cada latido.

Símbolo depulso➔ Sugerencia:
No hable ni se mueva durante las mediciones

AVG

Este símbolo aparece cuando la pantalla
LCD muestra el valor medio de las últimas 3
lecturas.

Promedio delas últimas 3
lecturas13SÍMBOLOS/
SÍNTOMASAVG
No se enciende
presionando el botón

AVG
AVG
El símbolo de error de

medición mostrado enla pantalla o el valor
de presión arterial semuestra excesivamente
alto o excesivamente
bajo.

CONDICIONES/CAUSAS

INDICACIÓN/CORRECCIÓN

Pilas agotadas.

Cámbielas por 2 pilas nuevas
alcalinas AAA (LR03).

La polaridad de las pilas se ha
colocado incorrectamente.

Vuelva a insertar las pilas
con la polaridad correcta.

La manga se ha colocado deforma incorrecta.

Coloque la manga
adecuadamente en su
posición correcta.

¿Habló o se movió durante la
medición?

Realice la medición denuevo. Mantenga la muñeca
firme durante la medición.

La muñeca se ha agitado
mientras el brazalete estaba
puesto.

Símbolo de error Anomalía en el circuito de aire.
El tubo del brazalete puede no
de medición
estar correctamente conectado
al tensiómetro.

Compruebe la conexión delbrazalete.
Realice la medición denuevo.

Símbolo de error La presión de inflado supera elvalor de 300 mmHg.
de medición

Apague la unidad y vuelva arealizar la medición.

Símbolo de error No se pueden determinarlos datos de medición de la
de medición
presión arterial.

Envuelva la mangacorrectamente y permanezca
quieto.
Realice la medición denuevo.

Nota: si "EP" aparece en la pantalla, devuelva el dispositivo a su distribuidor local oimportador.

149. ¿ Cómo medir ?

A. Insertar pilas en posición:a. Abrir la tapa de pila.
b. Referir a la foto por posiciones de las pilas. (Figura No. 4)
c. Todos los segmentos LCD aparecen en el monitor por 3 segundos.
B. Puesta de Fecha y Horas:a. Presione el botón " " durante más de 2 segundos ("año"comenzará a parpadear)
Presione el botón "" para establecer el año correcto (2011,2012,…2026)
b. Pulse el botón “ ” (“mes” comienza a parpadear)
Presione "" para programar el mes correcto (1, 2, 3,.....12).
c. Presione " " de nuevo (“fecha” empieza a parpadear)
Pulse el botón “” para establecer la fecha correcta
d. Presione " " de nuevo (“hora” empieza a parpadear)
Presione "" para programar la hora correcta en formato de 12
horase. Presione " " de nuevo (“minuto” empieza a parpadear)
Presione "" para programar el minuto correcto (0, 1, 2, 3,......
59)
f. Presione " " de nuevo para cambiar al tiempo normal(Mes, fecha, hora y minuto apareceran en el monitor)
C. Pasos para tomar la medición de presión arterial:a. Enrolle la manga sobre la muñeca. (Refiera a Enrolle la manga deltensiómetro en la muñeca como en Figura No. 5)
b. Sientese derecho en la silla para tener la posición correcta. (Refieraa Posición de medición correcta como Figura No.6)
15

c. Oprima "". Todos los elementos muestran “888” por 3
segundos. Luego, exhibe la hora exacta. El símbolo de pulso
parpadeará en la pantalla y, a continuación, la presión del aireaumentará automáticamente hasta 195 mmHg. Si la presión arterial
tiende a subir, la presión de aire aumenta en forma automática a onivel requerido para las mediciones.
*¡Nota!
No infle la manga hasta que esté envuelto en la muñeca.
* No se mueva o hable durante la medición de la presión.
d. Después de que la presión del aire aumente, disminuirá lentamente.
comenzará a parpadear.
Cuando se detecte el pulso,*¡Observación!
• Si la manga no para de inflarse, retírelo inmediatamente.
• Cuando se desee parar la medición, pulse "".
El aire de la manga saldrá y se detendrá la medición."e. Después de medir la presión arterial, la presión arterial sistólica,
la presión arterial diastólica y el pulso se mostrarán en la pantalla
durante 1 minuto.
D. Almacenar, recuperar y borrar de datos de medición:a. Almacenamiento de datos:
Después de cada medición de la presión arterial, la presión arterial
sistólica, la presión arterial diastólica, el pulso y la fecha y hora deldía específico se almacenarán automáticamente.
La memoria retiene los últimos 99 datos de medición. En caso deser más de 99 datos, la memoria borra los más antiguos en forma
automática.
16

b. Recuperación de datos:(1). Presione el botón " M ". Si no hay datos en la memoria,
solamente se mostrará HORA/FECHA en la pantalla. En elcaso de que haya datos, la primera lectura será el promediode las últimas 3 mediciones. (El valor medio no aparece si hay
2 mediciones o menos almacenadas en la memoria).
(2). Presione de nuevo el botón " M ". La última medición
aparecerá con el número así como la fecha y hora de
medición. Los datos que aparecen en la pantalla están
numerados correspondientemente. Tras leer los datos
recuperados como el conjunto de rotadores 99 de los datos
almacenados, pulse " M " para cambiar al tiempo normal.
" para tiempo(3). Para parar la lectura del dato, oprime "normal.
c. Borradura de datos:(1). Presione el botón " M ".
"y"" para borrar las(2). Presione sin soltar el botón "memorias. "CLA" aparecerá en la pantalla durante 3 segundos.
Nota: Una vez eliminados, los datos NO se podrán recuperar.
E. Cambio de pilas.
Cuando "" aparece en el display, la pila está baja.
No será posible tomar la medición de la presión arterial.
Todas las pilas utilizadas deben ser del mismo tipo. No mezcle
pilas alcalinas, normales (carbono-cinc) o recargables (cadmio).
No mezcle pilas nuevas con pilas usadas.
Siempre cambie las dos pilas.
17

10.Mantenimiento

Utilice un paño humedecido con agua o un agente de limpieza
medio para limpiar el dispositivo y séquelo inmediatamente
con un paño seco.
No utilice detergentes ni ningún producto químico para limpiar
el dispositivo.
Desinfección - Utilice un paño con una dilución de detergente
neutro o espuma para limpiar la superficie de la manga
durante 10 segundos.
Asegúrese de que la manga este completamente seca antes
volver a usarlo.
Si necesita algún consejo o servicio, consulte a su distribuidor
local.
No implemente los procedimientos de mantenimiento para eldispositivo durante la medición.
Solamente técnicos con la formación adecuada podrán reparary desmontar el dispositivo, incluidas las actualizaciones delsoftware, revisiones y mantenimiento.

*¡Nota!
Calidad del agua requerida para la limpieza y desinfección:
Agua corriente.

18

11.Medidas de seguridad

Guarde siempre el dispositivo en el estuche después deutilizarlo.
No exponga el equipo bajo la luz solar, alta temperatura,
humedad o lugares con mucho polvo.
No guarde el dispositivo en muy bajas (menos de –13 °F/ -25°C) y altas (más de 158 °F/ 70 °C) temperaturas, ni en un lugar
donde la humedad seá superior a 93% R.H.

12.Para prevenir el mal funcionamiento

No tuerza la manga de ninguna manera.
No inicie la medición cuando la manga no está envuelta en la
muñeca.
No intente a desarmar o cambiar nada del tensiometro,incluyendo la manga.
No deje caer el producto ni lo someta a fuertes impactos.

19

13.Advertencia:

El equipo sirve para medir la presión arterial en adulos
sólamente.
El caso de que la presión de aire en la manga no se detenga,
por favor abrir la maga inmediatamente.
Para garantizar la medición precisa se recomienda recalibrar
el dispositivo después de 2 años a partir de la fecha decompra.

Garantía
Este dispositivo tiene una garantía de 1 años desde la fechade compra. Si se encuentra un problema de fabricación yes necesaria su reparación, póngase en contacto con su
distribuidor.
La garantía no cubre los daños causados por: uso indebido
o abuso del producto, accidente, la inserción de cualquiermodificación no autorizada de los accesorios, instalación
inadecuada, reparación o modificación no autorizada, uso
inadecuado de la alimentación eléctrica/energía y por no
seguir el mantenimiento requerido.

20

14.Specofocations:
ModeloMétodo de medición
Alcance de medición: CH-617: Oscilométrico: Presión 0 ~ 300 mmHg,
Pulso 40 ~ 199 latidos/minuto
Precisión: Presión +/-3 mmHg
Pulso +/- 5% de lectura
Inflación: Bomba automática de aire
Deflación de presión: Liberación automática de aire con válvula
Pantalla: Pantalla digital de cristal líquido
Memoria: 99 mediciones
Dimensiones: Alto 70 x ancho 72 x grosor 31 mm
LA2,75 x AN2,83 x AL1,22 pulgadas
Peso: 88 ± 5 g (3.10 ± 0.17 oz) (pilas y manga excluido)
Tamaño de muñequera: Muñeca, circunferencia de 135 ~ 195 mm
Entorno de almacenamiento : Temperatura: -25 °C ~ 70 °C (-13 °F ~ 158 °F)
y transporte
Humedad: ≤ H.R. del 93 %Condiciones: Temperatura: 5 °C ~ 40 °C (41 °F ~ 104 °F)
medioambientales
Humedad: H.R. entre el 15 % y el 93 %Presión atmosférica: 700 hPa ~ 1060 hPa
Apagado automático
: Si no se utiliza la unidad durante 1 minuto
Vida útil del producto: 5 años (utilizándolo 4 veces al día)
Voltaje: 2 x Pila alcalina AAA LR03 (2 x 1,5 V) (3 VCC)
Duración de Pilas: Aproximadamente 250 mediciones
Accesorios: Estuche, manual de instrucciones, 2 pilas AAA LR03
Estas especificaciones están sujetas a cambio sin previo aviso con el fin demejorar el producto.

21

Nota:
Este monitor de presión arterial cumple la Directiva

0197 EC (93/42/EEC) y lleva la marca CE. Este

tensiómetro digital satisface también los requisitos delas siguientes normas predominantes (entre otras).

Norma de seguridad:
EN 60601-1
Equipos electromédicos, Parte 1: Requisitos generales para laseguridad básica y funcionamiento esencial.
Norma EMC:
EN 60601-1-2
Equipos electromédicos, Parte 1-2: Requisitos generales para laseguridad básica y características de funcionamiento esencial,Norma colateral: Perturbaciones electromagnéticas; Requisitos y
ensayos.
Normas de rendimiento:
EN1060-1
Esfigmomanómetros no invasivos - Parte 1: Requisitos generales.
EN 1060-3
Esfigmomanómetros no invasivos - Parte 3: Requisitos
suplementarios aplicables a los sistemas electromecánicos demedición de la presión sanguínea.
EN 1060-4
Esfigmomanómetros no invasivos - Parte 4: Procedimientos
de ensayo para determinar la exactitud del sistema total deesfigmomanómetros no invasivos automatizados.
EN ISO 81060-1 Esfigmomanómetros no invasivos - Parte 1: Requisitos y métodosde ensayo para el tipo de medida no automatizada.
IEC 80601-2-30 Equipos electromédicos - Parte 2-30: Requisitos particulares para laseguridad básica y funcionamiento esencial de los esfigmomanómetros automáticos
no invasivos.
Siga las instrucciones de uso.
22

Fecha de fabricación
PARTE APLICADA DE TIPO BF- Equipo con alimentación interna- No adecuado para utilizar en presencia de una mezcla
anestésica con aire, oxígeno u óxido nitroso- Operación continua con una carga de poca duración
Grados de protección que ofrecen las carcasas
Para evitar resultados incorrectos, causados porinterferencias electromagnéticas con equipos electrónicos,
no use este aparato cerca de un móvil o microondas.
Mantenga al menos una potencia de salida máxima de 2 W y
una distancia de separación de 3,3 m respecto a este equipo.
Deseche los productos usados según las normativas dereciclado locales.
Fabricante:
HEALTH & LIFE Co., Ltd.
9F, No. 186, Jian Yi Road, Zhonghe District,
New Taipei City, Taiwan
www.healthandlife.com.tw
Representante autorizado en la Comunidad
Europea:EMERGO EUROPE
Prinsessegracht 20, 2514 AP The Hague, The Netherlands23

Número de serie

LOT

Código de lote
Limitación de la presión atmosférica
Limitación de la humedad
Limitación de temperatura

MARCO DE RAEESi desea deshacerse de este producto, no lo mezcle con
residuos domésticos de carácter general. Existen unos sistemas
de recopilación independientes para productos electrónicos
utilizados conforme a la legislación bajo la directiva WEEE
(Directiva 2012/19/UE) y solamente se aplican dentro de la unión
europea.
Información para usuarios sobre la recogida y eliminación de pilasusadas.
El símbolo de esta hoja informativa significa que las pilas usadas
no se deben mezclar con los desechos domésticos generales.
Para llevar a cabo
un tratamiento, una recuperación y un
0197
reciclado adecuado de las pilas usadas, llévelas a los puntos dePbrecogida correspondientes.
Para obtener más información sobre la recogida y el reciclado de pilas,póngase en contacto con su ayuntamiento, el servicio de eliminación deresiduos local o el punto de venta donde adquirió los artículos.
24

0537

D

• CITIZEN is a registered trademark of Citizen Watch Co., Ltd. Japan.

6-1-12, Tanashi-cho, Nishi-Tokyo-shi,
Tokyo 188-8511, Japan
E-mail: sales-oe@systems.citizen.co.jp
http://www.citizen-systems.co.jp/

P/N:323103254

VER:001• Design and specifications are subject to change without notice.