English

FOR

DIGITAL BLOOD PRESSURE MONITOR

MODEL CH607
DAT
E

SE

E

Deutch

T IM

SY

T

CH
607
P

mm
Hg

DI

- Eng 1 -IN

EnglishEspañolPortuguês
DeutschFrançais

PUL
/M SE

ME
MO
RY

mm A .
Hg
STA
RT
/ ST
OP

MO
DE

mm S .
Hg

M

Español Português Italiano

INSTRUCTION MANUALGENERAL REMARKS
General remarks on blood pressure and blood pressure measurement• If the cuff is to be deflated quickly, depress “Start/
Stop” switch.
• The attached cuff is suited for a wrist circumference of 13.5 cm to 19.5 cm.
• When attaching the cuff, the body of the bloodpressure monitor should be placed on the inside of your wrist.
• Self-measurement is no therapy! In no caseshould the dosage of medication ordered by thedoctor be changed.
• In preparation for blood pressure measurement, you should urinate and should remain
relatively still for 10 to 15 minutes prior to measurement.
• Exercise, eating and drinking, smoking, etc.,
prior to measurement may affect the results.
• The blood pressure varies constantly throughout the day. Measurement should be maderegularly at the same time each day.
• Do not be too impressed by the results of one
measurement. Keep a record of blood pressure
variations. Many readings tell a story.

• Emotional stress may tend to cause blood pressure to rise.
• This product should not be used to monitor
cardiac arrhythmia.

For proper measurement of bloodpressure• Take a deep breath five or six times and thenrelax before measuring blood pressure. If you
are tense when taking the measurement, youwill not get a valid reading• Blood pressure will be elevated when you arein a worrisome or irritated state, suffering fromlack of sleep or constipation, or after exercise
or a meal.
• Putting on the cuff is one of the most important tasks for ensuring proper measurement ofblood pressure. Wind the cuff around the wrist
and tighten it so that one finger may be inserted
between the cuff and wrist.
• Do NOT measure blood pressure after bathingor drinking.
• When you feel like relieving yourself, you
should do so before measuring blood pressure.

- Eng 2 -• DO NOT measure blood pressure after the cuff
has been on for a few minutes, as your arm
will have built up higher blood pressure, andyou will not get a valid reading.

Precaution for use- Eng 3 -Français• Congestion, swelling, etc. may occur in somepeople.
• If blood pressure is measured repeatedly, an
accurate result will not be achieved. An interval of about 10 minutes should be allowed.
• The memory is erased if the batteries are removed for replacement, etc. and a record
should be kept of any necessary data.

Deutch• If you suffer form heart disease, high bloodpressure or other circulatory disease, please
consult your doctor first.
• If you feel abnormal pressure of the cuff during use or any other irregularity, reduce thepressure by immediately switching off thepower or removing the batteries and consult
the sale outlet where you purchased.
• If you think the measured value is abnormal or
if measuring makes you fell unwell, discontinue
use and consult your doctor.
• Measurement of blood pressure may not be
possible for someone with a weak pulse or arrhythmia.

Español Português Italiano

PRECAUTION FOR USE

English• Measure blood pressure where the room temperature is about 20°C. DO NOT measure bloodpressure when it is too cold (below 10°C)or too
hot (above 40°C) in the room .
• DO NOT try to measure blood pressure immediately after drinking coffee or tea or smoking.
• Measure blood pressure with a relaxed andquiet posture. Keep the center of the cuff at thelevel of your heart, and do NOT move your arm
or talk.

Precaution for maintenance

Cleaning method of Cuff

• Do NOT store the blood pressure meter where
it will be exposed to direct sunlight, high temperature (over 60°C), low temperature (below- 20°C), high relative humidity (over 95%) or
in a dusty place .
• Do NOT expose the blood pressure meter toundue force or vibration, or drop it.
• Remove the batteries when not use for a longtime period.
• Do NOT disassemble the apparatus.
• Do NOT bend the cuff or Air hose unnecessarily.
• Never wipe the blood pressure monitor withthinner or benzene, as they may damage it.
• When the monitor is very dirty, wipe it clean
with a cloth moistened with Sterilizing alcohol
or a neutral detergent, and then wipe it with adry cloth. Do NOT use thinner or benzene.

After cleaning the cuff with a neutral detergent,lightly rinse it with water and let it air dry. Hardrubbing or washing in a washing machine willcause air leakage. Also, be careful not to getwater into the air hose.

Protect the natural environment
Please help protect the natural environment by
respecting national and/or local recycling regulations when disposing of the batteries and the
products at the end of their useful lives.

Nature and frequency of maintenanceThis product is designed for use over an extended
period of time; however it is generally recommended that it be inspected every two years toensure proper function and performance.

- Eng 4 -

Reference European standard:

SYSTOLIC(mmHg)

This products complies with EN instructions 93/
42/EEC (Medical Device Directive) and with thefollowing standards: EN 1060:1995 “Noninvasive
Sphygmomanometers, Part 1 General Requirements” and EN 1060: 1997 “Noninvasive Sphygmomanometers, Part 3 Supplementary Requirements
for Electromechanical Blood Pressure Measuring
Systems”.

180

Hypertension (severe)
160
140
130

Manufacturer:

Hypertension (moderate)

Name: Living Science (Shenzhen) Co., Ltd.
Address: Shun Mu Ku Road, Ping Hu County,
Shenzhen, Long-Kang Town, China

Grade 1 Hypertension (mild)
High-Normal
NormalEuropean Representative:

120

80

85

90 100 110 DIASTOLIC

Name: Innovative business promotion GmbH.
Address: Huttenstraße 4 D-07743 Jena

(mmHg)

Note: When a patient’s systolic and diastolic bloodpressure fall into different categories, thehigher category should apply.

- Eng 5 -FrançaisDefinition and Classification of
Blood Pressure Lebels

DeutchOptimalEspañol Português Italiano

Standard for assessment of high or low blood pressure without regard to age, have been establishedby the World Health Organization (WHO), as shownin the chart.

EnglishWHO blood pressure classifications

DEVICE DESCRIPTION
1. Name of the parts
LCD display
DA
TE
CH
60
7

mm
Hg
P

M

TIM

S

mmY S
Hg .
ST
AR
T/

DI

ST
OP

mm A
Hg .
PU

M

O

D

E

T
E
S

E

/ M LSE
IN

Y
OR
EM
M

Setting button
Mode button
Recall memory button

Wrist cuff

Start/Stop button

Batteries“AAA” 1.5V×2

Carrying case

- Eng 6 -English

2. Description of display symbols
Hour Minute

Afternoon time
Measuring pressure
symbolSystolic rate

Blood pressure unit

Español Português Italiano

Month Date

Diastolic rate

Error symbol
Memory record

Pulse rate

- Eng 7 -Français

Pulse symbol

DeutchBattery weak displa y

BATTERY REPLACEMENT
1. To use this unit, you must install batteries, If
the battery mark “” is displayed, youmust change both batteries.
2. Open battery cover as illustrated. Replace thetwo batteries, matching the (+) and (–) polarities on each battery with the polarity markson the case.
• To take the batteries out easily, pull the transparent ribbon.

3. “LR03” batteries or equivalent are suitable forthis blood pressure monitor. All batteries usedmust be same type. Do not mix alkaline, standard (carbon-Zinc) or rechargeable (cadmium)batteries. Also, do not mix old and new batteries.

CLOCK MODE
1. Default time indicating state andselection of each mode
When the battery is installed, the time and dateare set as follows.
January 1 , 12:00:00 AM

2. Setting Date and Time
1) Press the “Mode” switch then “Month” starts
flashing and press “SET” button to set the correct month. (Each time the “Set” switch ispressed, the value of the month is increased- Eng 8 -

MEASUREMENT PROCEDURES- Eng 9 -Français

The blood pressure is measured at the wrist.
Therefore, rolling up the sleeve is no longer
necessary. The pressure cuff has integrated into
the wristlet. Wind the wristlet around your left
wrist. The wrist should be free. The display isat the inner of the wrist.

Deutch

1. Attaching of wrist cuff

Español Português Italiano

4) Press the “Mode” switch then “Minute” starts
flashing. Press “SET” button to set the correct Minute. (Each time the “Set” switch ispressed, the value of the minute is increased
by one. The value of Minute can be corrected
in 12-hour format in only the forward direction.)
5) Press the mode switch to finish correction.

Englishby one. The value of minute can be corrected
in only the forward direction.)
2) Press the “Mode” switch then “Date” starts
flashing. Press “SET” button to set the correct date. (Each time the “Set” switch ispressed, the value of the date is increased byone. The value of hour can be corrected inonly the forward direction.)
3) Press the “Mode” switch then “Hour” starts
flashing. Press “SET” button to set the correct date. (Each time the “Set” switch ispressed, the value of the Hour is increased byone. The value of hour can be corrected inonly the forward direction.)• Apply the wrist cuff to the wrist and make sure
it is not too loose or too tight. If the wrist cuff
is applied inappropriately, it may display an error or abnormal values. Pull the wristlet tightand close the “Burr closure”. Wrist circumference: 13.5-19.5 cm.

Measurement Procedures for
People With a Weak Pulse• Apply the middle part of the wrist cuff to theartery in the wrist to measure blood pressure.

2. To start the Measurement

CAUTION
Do not grasp the unit too tightly. Keep your left
wrist relaxed during measurement period. Do
not apply physical pressure against unit withwrist before or during measurement. All measurements must be performed while you areseated and with your left hand on a table held alevel with your heart. Keep your body still andrelaxed during measurement.

1) Press the “Start/Stop” switch. All display marks
appear for approx. 1 second. All displays disappear and the time is updated.
2) The blood pressure measuring symbol will appear on the display and then the air pressure
will automatically pump up to 195mmHg.
3) If blood pressure tends to be on the high level,
the air pressure will automatically increaseagain to the required level for measurement.

- Eng 10 -

4. Automatic power-off function
If the unit is left on after measurement, an automatic power-off function turns the power off after approximately 1 minute. To continue measurement, press the “START/STOP” switch turning the unit on. After, wind the wristlet aroundyour left wrist.

- Eng 11 -Français

To discontinue a measurement for any reason,press the “START/STOP” switch. This causes
the “wrist cuff” to deflate rapidly.

DeutchCAUTION

When the Start/Stop switch is pressed, the display will be changed over to the date/watch only
indication and the power will be totally turned
off approximately one minute later.

Español Português Italiano

3. Turn Power off

EnglishDo not move or talk while taking the blood pressure measurement.
4) When pressurization is completed, the automatic air release system will gradually reduce
cuff pressure, and measuring mark “Heartmark” will flash on the display indication thatmeasuring is in progress.
5) After taking blood pressure measurement, thesystolic and diastolic pressure with pulse rateis displayed on the display for 1 minute.
6) Unless an error is generated, the measurement
results are automatically stored in memory. Up
to 99 sets of results can be stored in memory.
* When there are already 99 sets of results inmemory, the oldest one is automatically erased
and overwritten with the newest. Display of themeasurement result will continue until power isturned off.

5. Memory recall

6. Memory clear function

You can recall stored data by pressing the“Memory” switch, except when performing ameasurement.
• Up to 99 sets of measurement results are automatically stored in memory, and memory cell
No. 1 is always used to store the oldest set ofresults.
• Press “Memory” switch, the memory cell number with the stored measurement results (Systolic / Diastolic and Pulse with Date/Time) aredisplayed for 1 minute.
* Pressing “Memory” button, nothing will appear
on the display if there is no data in the memory.
• If the “Memory” switch is pressed further, theold date are indicated in the order of the timeseries.
• Press “Start/Stop” switch to turn off the unit.

To clear memory data, take out the two batteries
at same time and then press the “START/STOP”
switch for a while. All memory is erased.

Press the “POWER/START” switch to turn off.

CAUTIONTo discontinue a measurement for any reason,press the “POWER/START” switch. This causes
the “wrist cuff” to deflate rapidly.

- Eng 12 -

DISPLAY MARK

Appears when the accurate bloodpressure could not be obtained accurately.

Press “Start/Stop” switch again andre-measure.
Or check cuff if wrapped at the wrist
or according to instructions.
Check palm if exerting effort.
Check if talking or moving during
measurement.
Check if posture is incorrect.

Replace batteries

Appears when the battery voltage isexcessively low or the positions ofbatteries are incorrect.

Replace all batteries with new ones.
Insert the batteries at paying close
attention to the +/– directions.

Measuring bloodpressure
symbolAppears all the time until the measurement is completed.

- Eng 13 -Français

Measuring Error

DeutchCORRECTIVE ACTION
Measurement in progress. Remain
still.

Español Português Italiano

CONDITION / CAUSEMark appears in the measurement
condition and flashes when pulse isdetected.

EnglishERROR DESCRIPTION OF DISPLAY MARKS

SPECIFICATIONS
Model Number
Measuring Principle
Measurement Localization
Cuff
Arm Circumference Range
Measurement Range
Accuracy

Pressure
Pulse
Pressure
Pulse
Pressure
Pulse

LCD Indication
Marks

LCD

Viewing Size
Digit Size

SwitchInflation
Deflation
ExhaustRated Voltage
Power Source
Power Consumption
Battery Life
Automatic Power Off Function
Clock Function
Memory Function
Dimensions
WeightOperating Environment
Storage Environment
Accessories

Temperature
Humidity
Temperature
Humidity

CH607
Oscillometric System
Left Wrist Arm
Preformed cuff
13.5 to 19.5 cm
0 - 300 mmHg
40 - 199 Pulse / Min.
±3mmHg±5% of reading
Systolic Blood Pressure: 3 digits
Diastolic Blood Pressure: 3 digits
Alternatively
Pulse Rate:
3 digits
Measuring
ExhaustRe-Measure
Battery38.0(H) × 32.0(W) mm
9.5(H) × 5.0(W) mm
4-switches (Start/Stop, Memory, Set, Mode)
Automatic Inflation by Air Pump
Automatic Air release control valve
Quick Exhaust Valve
3 V DC
2 “AAA” Size Batteries
0.1 W
Approx. 250 times measurement
Approx. 3 min. (after activated)±20 Seconds a month, Full Automatic Calendar, Alarm
99 Memory (Date, Time, Blood pressure value, Pulse)72(W) × 70(D) × 29(H)mm
Approx. 140g including batteries.
10 to 40°C
30 to 85%RH–20 to 50°C
10 to 95%RH
2 “AAA” batteries, Instruction manual and Carrying case

* Specifications and design may be changed without any notice in the event of improvementsbeing made.

- Eng 14 -