IMADA CO., LTD.
Tension Attachment Table

Tension Attachment TableThe suitable tension attachment varies depending on the shape of measurement sample.
We introduce the suitable tension attachment for each shape with classifying into 6 types.
Sample Classifications
1. Thin samples

3. Fine samples

5. Hooked samples

2. Slightly thick samples

4. Round bar/Columnar samples

6. Others[1. Thin Samples] Thin Samples such as films and rubber sheets
Features

Lift the lever and fix with one action

Wrap and fix (firm gripwith little damages tothe sample)

Fix with the lever orthe screw

ImagesMax.
SampleThickness

2mm

2 to 4mm

2mm

0.2 to 2mm

Capacity

1000N

500 to 2000N

500N

250 to 500N

Mounting

M6

M6 or M10

M6

M6

KC-1001

GR Series（GR-30/GR-2000）

GC Series（GC-60/GC-100）

FC Series（FC-21/FC-20/
FC-40/FC-21U/
FC-41U-FC-21UQ）

Model[2. Slightly Thick Samples] Slightly thick samples such as connectors and cables
Features

Lift the lever and fix with one action

Pinch grip with oneaction

Fix with the screw

ImagesMax.
SampleThickness

4.5 to 10mm

5 to 8mm

6 to 10mm

10 to 30mm

Capacity

250 to 1000N

500 to 5000N

15 to 20N

1000 to 5000N

Mounting

M6

M6 or M10

S-1/SR-1 (standardattachment) or M10

M6 or M10

ModelGP Series（GP-15/GP-30）

KC Series（KC-100/KC-5000N）PGC Series（PGC-0510/PGC2530）

GC Series（GC-1100/GC1200/GC-5000/GCF1200）2022/02/01

IMADA CO., LTD.
Tension Attachment Table[3. Fine samples] Fine samples such as threads and wires
Wrap and fix (firm gripwith little damages tothe sample)

Pinch grip with one
actionMax.
SampleDiameter

Φ1 to 3mm

Φ6mm

Φ0.5mmΦ1mm

Capacity

250 to 5000N

500N

50 to 100N

100 to 150NMounting

M6 or M10

S-1/SR-1(standard attachment)

Model

WG Series（WG-250N/WG1000N/ WG-5000N）

PGC-0505

FeaturesFix with the screw

ImagesM6FP Series（FP-50/51）

SC Series（SC-3/SC-8）[4. Round-Bar / Cylinder samples] Pen and bottle shaped samples
Features

For round-bar shaped samples

For cylinder shaped samples

ImagesMax.
SampleDiameter

Φ0 to 3.2mm

Φ0.5 to 13mm

Φ7 to 50mm

Capacity

150N

500 to 1000N

100N

Mounting
Model

M6
CP-150N

GC Series(GC-5/GC-15)

CCJ Series(CCJ-100N/CCJ-100N-01)[5. Hooked samples] Samples can be hooked such as cardboard and cap
FeaturesHook to open
For mold resin flip caps

Hook from both sides and pull
overcap and so on

Hook and pull cardboard andso on

50N

100N

250N

ImagesCapacity
Mounting
Model

M6
HFC-50N

SHF Series（SHF-100N/SHF-100N-L）

DF-60

2022/02/01

IMADA CO., LTD.
Tension Attachment Table[6. Others]
Use

For crimp terminal of wire harness test
For samples Terminal side
diameter of Φ0.2 to 6 mm

For samples Terminal side
diameter of Φ1 to 8 mm

For samples Terminal side
diameter of Φ4 to 12 mm

Model

CH-500N/CW-500N

FW-12/GP-30

CH-5000N/CW-5000N

Use

For polypropylene (PP)strapping band tension test

FeaturesImagesImagesUse

For loop tension test

ImagesModel

PPC-5000N

Model

LOP-1000N

Use

For QFP lead hook test

Use

For sewed buttons tension test

ImagesImagesModel

SJ-QFP-01

Model

BC-15/PGC-BC

Use

For can opening test

Use

For needle bonding test

ImagesModel

Images

PTD-100N/PTF-100N

Model

MED-NPT-200N

2022/02/01

IMADA CO., LTD.
Tension Attachment Table[Cautions]
▪
▪
▪
▪
▪

Information in this document is subject to change without prior notice.
The list of products listed intended for Force Measurements and not for any other purposes.
Unauthorized use and the reproduction without permission of the contents of this table prohibited.
This table is for product selection references and the guideline and does not guarantee various characteristics
or safety. Please refer to the individual specification for details.
Please note that the contents may not be applicable depending on the environment and conditions.

IMADA CO., LTD.
99, Jinnoshinden-Cho, Aza, Kanowari, Toyohashi,
Aichi 441-8077, JAPAN
Tel: +81-(0)532-33-3288
Fax: +81-(0)532-33-3866
E-mail: info@forcegauge.net
Website: http://www.forcegauge.net/en/

Visit our website for more
information on a wide rangeof product specifications,
measurement applicationsand videos.

2022/02/01