IMADA CO., LTD.
Long Stroke Type Manual Test Stand HV seriesLong Stroke Type Manual Test StandHV series
Ideal for measurement of larger or extensible samples due to the longer stroke
Allows stable measurement due to the handle operation

HV-500NⅡ

HV-1000N

Feature 1

Feature 2

Feature 3

Fine Movement

Various testing applications

Various options available

Allows fine adjustment of a force
gauge’s position and stable testingby the handle operation.

-Material strength tests (tensile,compress, shear strength, etc.)-Evaluation tests of mechanicaland electronic parts (solderingstrength of QFP, tensile strengthof crimp terminal, etc.)-Peeling tests of adhesive part andheat seals

Various options are available suchas length scale and long stroke fora wide range of measurement
needs.
*HV-3000N only have length scale
option (-S).

1 / 5
2022/03/11

IMADA CO., LTD.
Long Stroke Type Manual Test Stand HV series
Main options
Long stroke (-L) *1

Length meter (-S)

This option provides extended stroke.
Useful when testing stretching samples or testing forlong distance.

Built in digital screen indicates travel distance offorce gauge.

*1 HV-3000N is not available.

Specifications
Model

HV-500NⅡ

HV-1000N

HV-3000N

Capacity

500N

1000N

3000N

Trapezoidal screw withLoading Method

Ball screw with 4mm lead

3mm lead

Travel Rate

2mm/1 turn

1.2mm/1 turn

0.6mm/1 turnMax. Sample Height

Approx. 240mm

Approx. 375mm

Approx. 415mm

Table Material

Stainless Steel

Steel

Steel

WeightApprox. 10kg

Approx. 18kg

Approx. 24kg

Dimensions

See [Dimensions]Operation manual, tools

Accessories

GF-1(Grip mounting adapter)

Available Options-L: Extended stroke by200mm-S: Length scale

GF-2 (Grip mounting adapter)-L: Extended stroke by
150mm-S: Length scale-S: Length scale* The Max. Sample Height refers to the distance between the test stand table and the tip of force gauge (1000N or less ofZT series) measuring shaft when the head is at the top.

Mountable Force Gauges for HV series
DST/DSV series

ZTS/ZTA series

FB/PS/PSM series

Digital force gauge with excellent
cost performance capable tomanage data with PC.

Digital force gauge with highprecision, featuring high sampling
rate and a variety of usefulfunctions.

Mechanical force gauge for easy
handling.

2 / 52022/03/11

IMADA CO., LTD.
Long Stroke Type Manual Test Stand HV series[Dimensions]
HV-500NⅡ
Mounting Plate for Force Gauge

H
Length ScaleN

HV-500NⅡ

597.5

HV-500NⅡ-L

797.5

HV-500NⅡ-S

597.5

HV-500NⅡ-S-L

797.5

13.5

12.3

Unit: mm

3 / 52022/03/11

IMADA CO., LTD.
Long Stroke Type Manual Test Stand HV series
HV-1000N
Mounting Plate for Force Gauge

H

Length ScaleN

HV-1000N

730

HV-1000N-L

880

HV-1000N-S

730

HV-1000N-S-L

880

1513.8

Unit: mm

4 / 52022/03/11

IMADA CO., LTD.
Long Stroke Type Manual Test Stand HV series
HV-3000NA
HV-3000N25

HV-3000N-S

23.8

Length Scale

Unit: mm[Cautions]-Information in this document is subject to change without prior notice.
-This document is product descriptions and handling precautions, and do not guarantee various
characteristics or safety.
-This product is designed for force measurement purpose only.
-Do not copy and use this content without authorization.
-A force gauge (sold separately) is required to use this product.
-Do not use this product in the environments including fierce temperature changes, high temperature,high humidity, near water, dusty place

IMADA CO., LTD
99 Jinnoshinden-cho aza Kanowari Toyohashi
Japan 441-8077
Tel: +81-(0)532-33-3288
Fax: +81-(0)532-33-3866
E-mail: info@forcegauge.net
Website: http://www.forcegauge.net/en/

Visit our website for more
information on wide productspecifications, measurement
applications and videos.

5 / 52022/03/11