ＩＳＳＮ１３４６－７８４０

港湾空港技術研究所
資料
TECHNICAL NOTEOFTHE PORT AND AIRPORT RESEARCH INSTITUTE

No.1283

March 2014

港湾地域強震観測年報（2012）
野津
若井
長坂

独立行政法人厚淳陽介

港湾空港技術研究所

Independent Administrative Institution,

Port and Airport Research Institute, Japan目

次旨 ······························································································································································

3

1．はじめに ························································································································································

4

2．観測網と観測機器 ········································································································································

62.1

観測網 ····················································································································································

6

2.2

強震計の維持管理 ································································································································

72.3

観測地点 ················································································································································

7

2.4

強震計 ····················································································································································

7

2.5

強震計の基礎と観測小屋 ··················································································································· 11

要

3．記録の整理 ···················································································································································· 124．補正と積分 ···················································································································································· 13
4.1

フィルタリングと積分の方法 ··········································································································· 134.2計器特性補正等のためのフィルタ ··································································································· 14

4.3

SMAC-B2相当フィルタ ························································································································· 14

4.4

積分用のハイパスフィルタ ··············································································································· 15

4.5

解析のアウトプット ··························································································································· 165．2012年に取得された代表的な記録について ··························································································· 16
5.1

1月30日日向灘の地震（M4.9）の記録 ····························································································· 16

5.2

8月25日十勝地方南部の地震（M6.1）の記録 ················································································ 16

6．まとめ ···························································································································································· 19
謝辞 ······································································································································································· 19
参考文献 ······························································································································································ 20
強震観測担当者（2012年1月～12月） ··········································································································· 23-1-

Annual Report on Strong-Motion Earthquake
Records in Japanese Ports (2012)
Atsushi NOZU*
Atsushi WAKAI**
Yosuke NAGASAKA***

Synopsis
Since 1962, strong ground motions and earthquake responses of structures have been observed in
the major ports in Japan. In 2012, 752 accelerograms were obtained and analyzed at the Port andAirport Research Institute.
The strong-motion earthquake observation network in Japanese ports consists of 136
strong-motion accelerographs installed at 61 ports as of December 2012. Seventy-two
accelerographs out of 136 are installed on ground surface, 44 accelerographs are in ground by using
bore-hole and the rest 20 are on structures such as quay walls. Currently five types ofaccelerographs are used in the network, namely, the ERS accelerograph, the SMAC-MDU
accelerograph, the Datol-100 accelerograph, the Omni accelerograph and the Basalt accelerograph,
all of which are negative feedback accelerographs equipped with a digital recorder.
This report presents the results of observation and preliminary analysis of records obtained in2012. ASCII data of the recorded accelerations for all the records in 2012 are available on theattached DVD in the user-friendly CSV format. Also, computer plots of recorded accelerations areavailable on the DVD for the records with peak accelerations exceeding 20 Gal. For the records
with peak accelerations exceeding 50 Gal, computer plots of recorded accelerations, integrated
velocities and displacements, Fourier spectra and response spectra are available on the DVD.
Key Words: earthquake, port, strong-motion earthquake observation, acceleration record,
Fourier spectra* Head of Engineering Seismology Group, Earthquake Disaster Prevention Engineering Division** Postdoctoral Fellow, Engineering Seismology Group, Earthquake Disaster Prevention Engineering Division*** Research Engineer, Engineering Seismology Group, Earthquake Disaster Prevention Engineering Division
3-1-1 Nagase, Yokosuka, 239-0826 Japan
Phone：+81-46-844-5058
Fax：+81-46-844-0839
e-mail:nozu@pari.go.jp-2-港湾地域強震観測年報（2012）
野津

厚*・若井

淳**・長坂

要

陽介***

旨

1962 年より実施されている港湾地域強震観測により，2012 年には 752 の強震記録が得られた．強
震計の台数としては，2012 年末現在，136 台の強震計が 61 港に設置されており，設置状況としては，
72 台が地表に，44 台が地中に，20 台が構造物上に設置されている．観測網を構成する強震計は ERS型強震計，SMAC-MDU 型強震計，Datol-100 型強震計，Omni 型強震計，Basalt 型強震計の 5 種類であり，いずれもサーボ型加速度計である．
本資料は 2012 年に港湾地域強震観測網で得られた記録について報告するものである．得られた記録は港湾毎に分類され，地震諸元・記録番号・最大加速度等とともに付録 DVD の強震観測表にまとめられている．強震観測表の地震諸元は気象庁の地震・火山月報（カタログ編）に基づいている．
また，強震記録の記録番号は各観測地点から記録が送られてきた順番に付けられており，記録の分類として，頭文字「F」で始まる記録は ERS 型強震計で得られた記録，頭文字「U」で始まる記録はSMAC-MDU 型強震計で得られた記録，頭文字「D」で始まる記録は Datol-100 型強震計で得られた記録，頭文字「O」で始まる記録は Omni 型強震計で得られた記録，頭文字「B」で始まる記録は Basalt型強震計で得られた記録をそれぞれ示している．強震観測表にまとめられているすべての記録のデ
ジタルデータを付録 DVD に CSV 形式で収録している．最大加速度が 20Gal を越える記録について
は加速度波形を画像ファイルの形で付録 DVD に収録している．また，最大加速度が 50Gal を越える
比較的大きな記録については，加速度波形に加え，速度波形，変位波形，フーリエスペクトル，応答スペクトルも画像ファイルの形で付録 DVD に収録している．
2012 年の観測は次に示す諸機関の協力の下に実施された．
国土交通省港湾局
東京都港湾局

国土交通省地方整備局

静岡県港湾総室

国土交通省北海道開発局

大阪市港湾局

神戸市みなと総局

内閣府沖縄総合事務局
宮崎県港湾課

強震観測は，各観測地点での強震観測担当者の努力に負うところが非常に大きい．担当者各位に敬意と謝意を表すとともに，各観測地点で実際に観測に携わった方々の氏名を巻末に掲載する．
キーワード：地震，港湾地域強震観測，加速度記録，フーリエスペクトル* 地震防災研究領域 地震動研究チームリーダー** 地震防災研究領域 地震動研究チーム 特別研究員*** 地震防災研究領域 地震動研究チーム 研究官〒239-0826 横須賀市長瀬3-1-1 港湾空港技術研究所
電話：046-844-5058 Fax：046-844-0839
e-mail:nozu@pari.go.jp-3-

1. はじめに中小地震の記録を取得して，将来の大地震の揺れの予測に活用することである．上述のように地震動の性質は地

港湾地域強震観測は当所の前身である運輸省港湾技術点毎に異なるため，構造物の耐震設計には地点毎の特性研究所が中心となり 1962 年に開始され，国の機関や地方を反映した入力地震動を用いることが望ましい．各港湾

自治体が参画して実施されてきた．2012 年末の時点ではで得られた中小地震の記録には，港湾毎の地下構造の情

全国 61 の港に 136 台の強震計が設置されている（図-1）．

報が含まれており，この情報を生かすことにより，将来

図-1 には強震計の機種も示されているが，その詳細につの大地震による揺れの予測に役立てることができる．そいては 2.を参照されたい．表-1 に 2001 年以降の強震計の一例として，2000 年鳥取県西部地震の余震の際に境港台数と記録数の推移を示す．

とその周辺の複数の地点（図-3）で取得された記録を経験的グリーン関数法と呼ばれる方法で多数重ね合わせる

港湾地域強震観測の目的は極めて多岐にわたるが，ここでは代表的な三つの目的を説明する．

ことにより，鳥取県西部地震の本震地動をシミュレーシ（1）強震観測の第一の目的は，港湾に被害をもたらすよョンした結果を図-4 に示すが，各地点の地震動の振幅，

うな大地震が発生した場合に，その揺れを記録して，こ周期，継続時間などの特徴をかなり良好に再現できていれを被害メカニズムの解明や適切な復旧工法の選定に利ることがわかる．ここで用いた手法の詳細については文用することである．1995 年兵庫県南部地震で被災した神献 6）を参照していただきたい．
港湾地域強震観測で得られた記録は，後に説明するよ戸港の岸壁の被害原因の解明には，神戸港で取得された
1），2）．また，2011 年東北地方太平

うな処理と解析を経た後に，強震観測年報として公表さ洋沖地震の際に取得された強震記録も，防波堤等の被害れてきた．まず，1963 年から 1975 年の記録が強震観測

原因の解明に活用されている．

年報として公表された 7)-17)．1963 年から 1975 年の年報

強震記録が活用された（2）強震観測の第二の目的は，地点毎に異なる地震動のには鉛直成分が含まれていなかったので，それらは別冊

特性を解明することである．図-2 では八戸港と関西国際18)

空港で得られた強震記録のフーリエスペクトルを比較しからは新しいデータ処理の方法を取り入れ，計器補正をている．八戸港では 1968 年十勝沖地震と 1994 年三陸は施した加速度波形とそれを積分した速度波形と変位波形，

るか沖地震の強震記録が得られているが，26 年の時を隔

フーリエスペクトルや応答スペクトルを年報で報告するてて発生した二つの大地震で，いずれも周期 2.5 秒（周ようになった 19)-49)．また，通常の年報とは別に，被害地

波数 0.4Hz）の成分が卓越している．一方，関西国際空震の記録だけを集めた資料も刊行されている 50)-61)．強震

港では 1995 年兵庫県南部地震と 2000 年鳥取県西部地震

観測年報の変遷を表-2 に示す．

として公表された．1976 年と 1977 年の強震観測年報の記録が得られているが，いずれも周期 5 秒（周波数

昨今のように公的機関の組織再編が進む中では，強震

0.2Hz）の成分が卓越している．このように，場所により記録のデジタルデータを研究室内で保管することが最も地震動の特性が異なるのは，地震基盤上に存在する堆積

確実な保管方法であるとはもはや言えなくなってきてい層の地震動への影響（サイト特性）が場所毎に異なるたる．デジタルデータを保管する場所としてはウェブ・サめである．平成 19 年 4 月に改訂された新しい港湾の施設

ーバも考えられるが，ウェブ・サーバのサービスが停止の技術上の基準では，設計用入力地震動の評価を行う際，すれば，それと同時にデータが失われる可能性がある．

対象施設設置位置もしくはその周辺における地震動の実

しかるべき出版物に電子媒体を添付して，その中にデジ

測値に基づいてサイト特性の評価を行うことが求められ

タルデータを収めておくことが，強震記録のデジタルデるようになった．従って，設計実務において強震観測のータを後世に残すための最も確実な手段であろう．電子

果たす役割の重要性は今後一層高まるものと予想される．

媒体にも種々のものがあるが，容量・価格・普及の度合

このことに関連して，全国の港湾等の強震観測地点においなどから CD または DVD が最適と判断される．このけるサイト増幅特性の評価結果をとりまとめた資料が刊ような考えから 1999 年以降の強震観測年報 38)-49)には，

行されているので 3），4），5），参考にしていただければ幸い

デジタルデータを収録した CD または DVD を添付してである．強震観測により地震動の卓越周期の解明が進めいる．

ば，構造物の固有周期と地震動の卓越周期が一致しない

本資料の付録 DVD には，2012 年に取得されたすべてように工夫することも可能になるので，地震による被害の加速度記録のデジタルデータが CSV 形式で収められの軽減に資するものと期待される．

ている．また，20Gal を越える記録については，加速度（3）強震観測の第三の目的は，良質な（SN 比の高い）記録を図化したものが収められている．さらに，50Gal-4-

図-1 港湾地域強震観測網-5-

表-1 強震計台数と記録数の推移（台数は年末の数字）

年 港湾数 強震計台数 記録数
2001
60
97
207
2002
60
98
140
2003
60
104
394
2004
60
110
468
2005
60
111
434
2006
61
113
276
2007
61
119
520
2008
61
119
417
2009
61
119
488
2010
61
119
296
2011
61
136
2417
2012
61
136
752
表-2 強震観測年報の変遷

年1963～1975

強震観測年報の内容
CD
加速度波形，フーリエスペクトル，応 無
答スペクトル．鉛直成分は別冊として公表．
1976～1998
計器補正を施した加速度波形とそ 無れを積分した速度波形と変位波形，
フーリエスペクトル，応答スペクト
1999～本資料 同上有

図-2 八戸港と関西国際空港の地震動特性の比較

を越える記録については，加速度記録と，これを積分して得た速度波形，変位波形，フーリエスペクトル，さらに応答スペクトルを図化したものが収められている．

2. 観測網と観測機器

図-3 境港およびその周辺の強震観測地点

2.1

観測網

港湾地域強震観測網は日本全国の海岸線をカバーしており，2012 年末の時点では 136 の強震計が 61 の港に設置されている．2012 年末の時点で強震計の設置されている港湾
は図-1 および付録 DVD の地図に示すとおりである．付録
DVD の地図上の港名をクリックすれば，強震計の機種と設置条件が表示される．観測地点の土質柱状図等も付録 DVDに収められており，地図からたどることができる．観測地
点のさらなる詳細情報については観測地点資料 62)-67)を参照されたい．2012 年末の時点では，136 の観測地点のうち
67 地点に ERS 型強震計が，52 地点に SMAC-MDU 型強震計が，8 地点に Datol-100 型強震計が，3 地点に Omni 型強震計が，6 地点に Basalt 型強震計が設置されている．これらはいずれもサーボ型加速度計（フォースバランス型または速度帰還型）68）である．以前に主力であったアナログ式

図-4 経験的グリーン関数法による 2000 年鳥取県西部地
震の本震地動の再現 6）の SMAC-B2 型強震計は姿を消している．

-6-

2.2

強震計の維持管理

強震計の維持管理は当所と港湾地域強震観測の他の参画機関との緊密な協力の下に実地されている．現在，観測網を構成する強震計の大半は通信機能を有しており，ダイア
ルアップ方式により，観測点から当所へのデータ転送が可能な状況となっている．オンラインによるデータ転送が可能な地点は，2012 年末の時点では，136 の強震観測地点のうち 128 地点である．
強震計の通信機能は，データ収集だけでなく，強震計の健全度把握にも用いられている．いつ発生するかわからな
い大地震に備えて，強震計を常に健全な状態に保つことが極めて重要である．そのためには，強震計に不具合が発生した場合，それを早期に発見することが必要である．強震計の通信機能を活用し，当所の職員が強震計の健全度把握を行うことが，観測網の維持に大きく寄与していると考え

図-5 観測地点名のルール

られる．
なお，通信方式については，従来はダイアルアップ方式

観測地点の位置を表している．港湾事務所に設置されていであったが，地震直後に強震計から当所サーバまで波形デる観測点では「小名浜事」のように「事」という文字をつータを自動で転送できる新しいシステム（地震動情報即時けている場合がある．添え字の「U」は強震計の型を表し

伝達システム）が 2011 年に開発され，2012 年末の時点でており，釧路港には SMAC-MDU 型強震計が設置されていは 44 地点に導入されている．
この新しいシステムについてることを意味する．SMAC-B2 型強震計には添え字「S」が，は稿を改めて説明することとしたい．

ERS-B，C，D 型の強震計には添え字「M」が，ERS-F 型強
震計には添え字「F」が，ERS-G/GV 型強震計には添え字「G」

2.3

観測地点

が，SMAC-MDU 型強震計には添え字「U」が，Datol-100

観測網には 3 種類の観測地点がある．1 つ目は地表面の型強震計には添え字「D」が，Omni 型強震計には添え字「O」

加速度を記録するもの，2 つ目はボアホールを使用し地中が，Basalt 型強震計には添え字「B」が，それぞれ対応する．

の加速度を記録するもの，3 つ目は構造物の地震応答を記「釧路-UB」で添え字「U」の後ろの「B」は設置条件を示録するものである．構造物の地震応答を記録するための観し，ボアホールを使用して地中に強震計を設置しているこ

測点のそばには必ず地盤の加速度を測定するための観測点とを意味する．例えば「釧路-U」のように設置条件を表すが設けられている．

添え字がない場合は，その強震計が地表に設置されている各観測地点では，原則として，強震計の水平成分の一つことを意味する．添え字「R」がついている場合は強震計

は真北を向いている．ただし，例外として，構造物の応答が構造物上に設置されていることを意味する．「神戸」等のを記録する観測点と，これに付随して地盤加速度を記録す

ように地中の異なる深さに強震計が設置されている場合は，る観測点では，強震計が構造物と平行に設置されている場

添え字「B」の後ろに強震計の深度を表す「40」等の数字合もある．岸壁や桟橋など多くの港湾構造物は 2 次元的なをつけている（単位 m）．また，「清水日の出-UC1」と「清形状をしており，構造物の法線方向に垂直な成分と平行な

水日の出-UC2」の「C」は，強震計が CDM 改良地盤に設成分を計測する方が解析上便利であるため，そのようにし置されていることを示している．

ている．強震計の 2 つの水平成分が真の南北方向と東西方向に一致しない場合には，方向成分を表すために偏角（単2.4位は度）が使用される．例えば N10E は真北から東に 10°(1) ERS-F 型強震計

強震計

ERS-F 型強震計は不揮発性磁気バブルメモリを備えたサ

回転した方向を意味する．
観測網のそれぞれの観測地点には観測地点名が付けられーボ型加速度計（フォースバランス型）である．記録器のている．観測地点名は位置と強震計の型と設置条件の組み

メインユニットは，図-6 に示す通り，不揮発性磁気バブル

合わせである（図-5）．
例えば北海道の釧路港の観測点は「釧メモリが 4 つと制御装置からなり，その寸法は

路-U」および「釧路-UB」と名付けられている．「釧路」は240×240×35mm，質量は約 1kg である．記録器には 2 つの

-7-

表-3

ERS-F, G, GV 型強震計の仕様

総合特性(-F)2G
計測可能加速度 (-G, -GV)0.008G～2G の範囲で設定可
能
周波数範囲

0.01Hz-35Hz

ダイナミックレ
86dB over
ンジ
換振器
成分

図-6

水平 2 成分と上下成分(-F) 2G
計測可能加速度 (-G, -GV)0.008G～2G の範囲で設定可能ERS-F 型強震計の記録装置のメインユニット

感度

10-5

形式

(-F and -G)フォースバランス型(-GV)速度帰還型

フィルタ
ハイパスフィル
0.007Hz -6dB/octave
タ
ローパスフィル
35Hz -18dB/octave
タA/D 変換
解像度

16bit

変換速度

100Hz

遅延メモリ： 10s
図-7

時計: 内蔵時計の精度は 1/100s，1 時間毎にラジオの時報により校正ERS-F, G, GV 型強震計の地中設置用換振器

スタータ: トリガーレベルは計測可能加速度の 0.5，1
または 2％
記録器

図-8

ERS-F, G, GV 型強震計の構造物取り付け用換振器

チャンネル数(-F) 4-10 チャンネル (うち 1 チャン
ネルは時刻)(-G, -GV)3-12 チャンネル

メモリサイズ(-F) バブルメモリ: 1 Mbyte×2(-G, -GV) IC カードメモリ: 2 Mbyte

収録時間

1 記録あたり 1-10min

最大記録時間(-F) 40 min/4ch(-G, -GV) 52 min/3ch
加速度が大きいものを優先的に保存

関連情報

観測地点, 記録数,記録開始時刻, 各
成分の最大値ユニットが設置できる．加速度 3 成分に時間信号を加えた

校正：総合的な校正が可能

4 チャンネルの記録時間はユニットが 2 つの場合には約 40

バックアップ電源: 充電後 2 時間分である．

容器：アルミニウム箱，防水ERS-F 型強震計には，もともと換振器と記録器を 1 つのサ イ ズ ： (-F) 54(L)×54(W)×38(H)cm
54(L)×54(W)×33(H)cm

箱に収納した一体型のものがあり，その後，換振器と記録器が分かれた分離タイプのものが開発された．図-7 に示す

-8-

(-G,

-GV)

図-9

図-10

ERS-F, G 型強震計の周波数特性

図-11

ERS-GV 型強震計の周波数特性

ERS-G/GV 型強震計の IC カードメモリ

ものはボアホール用の換振器である．図-8 に示すものは構
造物取り付け用の換振器である．2012 年末の時点で ERS-F
型強震計の総数は 3 である．

図-12

ERS-G/GV 型強震計の外観

ERS-F 型強震計の仕様を表-3 に示す．ERS-F 型強震計の周波数特性を図-9 に示す．

（メモリサイズは 2MB）である．加速度 3 成分の記録時間(2) ERS-G/GV 型強震計

は 1 つのカードで約 52 分である．
加速度のトリガーレベル

ERS-G/GV 型強震計は記憶媒体として IC カードメモリをは様々な段階に設定できる．強震計の計測可能最大加速度

用いたサーボ型加速度計であり，フォースバランス型

は 2G で最大加速度のレベル（感度）は 0.008G から 2G の

（ERS-G 型）と速度帰還型（ERS-GV 型）がある．図-10間で適切な値に設定できる．2012 年末の時点で ERS-G/GV

は ERS-G/GV 型強震計で用いられている IC カードメモリ型強震計の総数は 64 である．

-9-

表-4

SMAC-MDU 型強震計の仕様（港湾地域強震観測での設定）
計測
計測範囲±2097Gal

振動数範囲

DC～30Hz

ダイナミック114dB
レンジ
A/D 変換解像度

24bit

サンプリング
100Hz
周波数
換振器
成分

水平 2 成分と上下成分

感度

3V/G

形式

フォースバランス型
図-13

フィルタ

SMAC-MDU 型強震計の外観

ハイパスフィ
0.02Hz
ルタ
ローパスフィ
30Hz
ルタ
時計
水晶精度

誤差±0.5ppm 以下

時刻修正

GPS による時刻修正

記録
起動条件

指定 3 成分の論理演算組み合わせ

記録開始

正秒記録開始

記録停止

停止設定レベルを下回って設定時間
後に自動停止最大記録時間600 s/3ch
加速度が大きいものを優先的に保存

メモリ

IC カードメモリ: 10Mbyte 以上

記録内容

加速度波形データ，起動時刻
設定値，機器状態，最大加速度値
SI 値，計測震度値

電源
使用電源

AC100V

50/60Hz

内蔵充電池

DC12V
バックアップ: 3 時間以上

図-14

SMAC-MDU 型強震計の周波数特性

外形寸法・質量
サイズ(据置型) 398W×418D×210Hmm，約 24kg型と異なっており，これを図-11 に示す．速度帰還型は雷の影響に対して有利であることから 68），ERS-GV 型強震計

ERS-G/GV 型強震計の仕様を表-3 に ERS-F 型と共に示す．

は地中観測点とこれに付随する地表観測点（例えば「秋田フォースバランス型である ERS-G 型強震計の周波数特性-GB」と「秋田-G」）で多く用いられている．換振器と記録

は ERS-F 型強震計と同じであり，図-9 に示すとおりである．

器を 1 つの箱に収納した標準的な ERS-G/GV 型強震計の外速度帰還型である ERS-GV 型強震計の周波数特性は ERS-F

観を図-12 に示す．

- 10 -

Datol-100 型強震計の仕様

表-5

表-6

Omni 型強震計の仕様（港湾地域強震観測での設定）（港湾地域強震観測での設定）
計測

計測
計測範囲

±1000Gal

計測範囲±2000Gal

振動数範囲

0.1～30Hz

振動数範囲

0.1～30Hz

A/D 変換解像度

A/D 変換
16bit

解像度

24bit

サンプリング
100Hz
周波数サンプリング
100Hz
周波数

換振器

換振器

成分

水平 2 成分と上下成分

成分

水平 2 成分と上下成分

感度10mV/Gal

感度

5mV/Gal

形式

速度帰還型

形式

速度帰還型

フィルタ

フィルタ

ローパスフィ
30Hz
ルタ

ローパスフィ
30Hz
ルタ

時計

時計

時刻修正

ラジオの時報により校正

時刻修正

記録

GPS による時刻修正

記録

起動条件

3 成分の論理演算組み合わせ

起動条件

3 成分の論理演算組み合わせ

記録停止

停止設定レベルを下回って設定時間
後に自動停止記録停止

停止設定レベルを下回って設定時間
後に自動停止保存モード

加速度が大きいものを優先的に保存

保存モード

新しいものを優先的に保存

メモリ

IC カードメモリ: 4～8Mbyte

メモリ

Flash Card: 256Mbyte

電源

電源

使用電源

AC100V

50/60Hz

内蔵充電池

DC12V
バックアップ: 約 5 時間

使用電源

AC100V

50/60Hz

内蔵充電池

バックアップ: 4 時間以上ある．2012 年末の時点で Omni 型強震計の総数は 3 である．
(3) SMAC-MDU 型強震計

(6) Basalt 型強震計

SMAC-MDU 型強震計はフォースバランス型のサーボ型

Basalt 型強震計はフォースバランス型のサーボ型加速度加速度計である．表-4 に SMAC-MDU 型強震計の仕様を示計である．表-7 に Basalt 型強震計の仕様を示す．強震計のす．強震計の外観と周波数特性をそれぞれ図-13 と図-14周波数特性を図-16 に示す．2012 年末の時点で Basalt 型強に示す．2012 年末の時点で SMAC-MDU 型強震計の総数は震計の総数は 6 である．

52 である．
(4) Datol-100 型強震計

2.5 強震計の基礎と観測小屋

Datol-100 型強震計は速度帰還型のサーボ型加速度計で港湾地域強震観測で強震計を地表に設置する場合，強震ある．表-5 に Datol-100 型強震計の仕様を示す．強震計の観測小屋に設置する場合とハンドホール内に設置する場合

周波数特性を図-15 に示す．2012 年末の時点で Datol-100

がある．このうち強震観測小屋に設置する場合，図-17 に型強震計の総数は 8 である．

示すような鉄筋コンクリート製の簡単な基礎を使用するこ(5) Omni 型強震計とが標準となっている．図-17 に示される厚さ 300mm の空

Omni 型強震計は速度帰還型のサーボ型加速度計である．

洞は，基礎のみかけの密度を周囲の土の密度と等しくする

表-6 に Omni 型強震計の仕様を示す．強震計の周波数特性

ことにより，基礎が存在することによる記録への影響を最は Datol-100 型強震計と同じであり，図-15 に示すとおりで小限とするために設けられている．基礎は強震計を収納す- 11 -

表-7

Basalt 型強震計の仕様（港湾地域強震観測での設定）
計測
計測範囲±3920Gal

振動数範囲

DC～40HzA/D 変換解像度

24bit

サンプリング
100Hz
周波数
換振器
成分

水平 2 成分と上下成分

感度1.276mV/Gal

形式

フォースバランス型

フィルタ
ローパスフィ
50Hz
ルタ
時計
時刻修正
図-15

GPS による時刻修正

記録

Datol-100/Omni 型強震計の周波数特性

起動条件

3 成分の論理演算組み合わせ

記録停止

停止設定レベルを下回って設定時間
後に自動停止（最小収録時間 90s）

保存モード

新しいものを優先的に保存

メモリ

CF カード 2.8GB

電源
使用電源

AC100-250V

50/60Hz

内蔵充電池

バックアップ: 約 30 時間

合には，コンクリートや木製の杭が使用されることもある．
例えば新潟港の観測点では基礎を杭で支持している．強震観測小屋そのものの構造には特に制限はないが，耐震性に優れたものである必要がある．
強震観測小屋の一例として，
小名浜港の強震観測小屋を図-18 に示す．

3. 記録の整理
当所に到着したすべての強震記録は以下に述べる手順
で整理・保管される．まず，各々の強震記録は，研究室に到着した順に番号がつけられる．ERS-F,G,GV 型強震計の記
図-16

Basalt 型強震計の周波数特性

録には「F」で始まる番号，SMAC-MDU 型強震計の記録には「U」で始まる番号，Datol-100 型強震計の記録には「D」

る小屋とは構造的に離れている必要がある．通常，強震計

で始まる番号，Omni 型強震計の記録には「O」で始まる番の基礎を支持するのに杭は使用されない．しかし，観測地

号，Basalt 型強震計の記録には「B」で始まる番号が付けら点の地盤が非常に軟弱な粘土地盤やゆるい砂地盤である場

れる．次に，各々の強震記録に対応する地震を確認する．

- 12 -

図-17

強震計基礎の一例

図-19

補正と積分の手順定められた方法 71), 72)に従い，計器特性補正のためのフィ
ルタリング，高周波成分と低周波成分の補正のためのフィ
ルタリング，積分，応答スペクトルとフーリエスペクトルの計算などを実施する．処理の手順を図-19 に示す．この
手順は 1976 年以降の記録に適用されてきている．図-19 においてオリジナル加速度波形とは基線補正のみを行った波形，補正加速度波形とは，計器特性に関する補正を行った
上で，SN 比の十分でない低周波成分をハイパスフィルタ
図-18

により削除した波形，SMAC-B2 相当加速度波形とは，港観測小屋の一例(小名浜事-G,GB)

湾地域強震観測で長く用いられてきた SMAC-B2 型強震計
による記録と比較可能なように周波数成分を調整した波形記録に対応する地震の震源情報としては，気象庁地震・火69）

山月報（防災編）のことである．

のものを仮に採用しておき，強震観測

年報の刊行までには気象庁地震・火山月報（カタログ編）
70）のものに置き換える．なお，付録 DVD の強震記録一覧

4.1 フィルタリングと積分の方法
フィルタリングと積分は周波数領域で実施する．リンクのマグニチュードは気象庁マグニチュードである．

効果を避けるため，基線補正後の加速度波形（オリジナル

4. 補正と積分

加速度波形）の外側にゼロを付けたものに対して FFT を適用する．ゼロ部分の長さ L(s)は次式により定める．

得られた記録に対して，まず基線補正が行われる．次に，- 13 -

L > max [2 / 3T,10.0]

(1)

図-20

SMAC-B2 相当フィルタ

図-21

固定フィルタと二回積分の合成

図-22

パラメタ付きフィルタと二回積分

ここに T(s)はオリジナル加速度波形の長さである．この条
件は，後述する積分用のハイパスフィルタのインパルス応
答を考慮して定めたものである．
4.2 計器特性補正等のためのフィルタ
計器特性補正としては，図-9，図-11，図-14，図-15 および図-16 の下段に示す位相特性に関する補正を行う．これに加え，Datol-100 型強震計および Omni 型強震計については，振幅特性の補正の必要性が高いので，図-15 の上段に示す振幅特性の補正も行う．さらに，すべての強震計に対して，振幅特性に関し，高周波成分を次式に示すフィル
タにより取り除く．
Af f

=

{

1

| f | ≤ f1

1 [cos (π( f – f )/( f – f ))+1]
1
2
1
2
0

f1≤| f | ≤ f2
f2≤| f |(2)
ここに f1=25Hz， f2=40Hz である．
4.3

SMAC-B2 相当フィルタ

港湾地域強震観測で長く用いられてきた SMAC-B2 型強震計の周波数特性は，現在用いられている強震計の周波数特性とは異なるので，現在の強震計で得られた記録を
SMAC-B2 型強震計による記録と比較できるようにするため，「SMAC-B2 相当加速度」を求める．そのため，次式に示すフィルタを用いる．

S (f) =
1–

図-23

f
fs

2

(3)

1
+ 2h sパラメタ付きフィルタと二回積分の合成

fifs

する．このフィルタを作用させることによって得られた「SMAC-B2 相当加速度」は，強震計で測定しようとしてここに fs=1/0.14=7.1Hz，hs=1.0 である．

いる地盤や構造物の本来の加速度とは異なっており，そのこのフィルタは図-20 に示されるとおり SMAC-B2 型強

最大値は本来の加速度よりも小さいが，この処理を施して震計と同じ周波数特性を有する．このフィルタは，4.2 で得られる加速度はすべて互いに比較可能である．

述べた計器特性補正のためのフィルタを適用した後に適用- 14 -

4.4 積分用のハイパスフィルタ
すべての周波数に対して SN 比を一定以上に保つと同時に物理的に意味のある信号はできるだけ消さないようにするという観点からは，積分用のハイパスフィルタの遮断周波数が加速度波形の周波数特性に応じて変化することが望ましい．一方，積分後に得られる速度波形や変位波形を異なる記録について相互に比較するという観点からは，遮断周波数が一定値であるほうが都合がよい．強震記録が様々な立場で応用されることを考慮し，ここでは，速度波形や変位波形を得るための積分方法として二通りの方法を用いる．一つは固定フィルタを用いる方法であり，もう一つはパラメタ付きフィルタを用いる方法である．
(1) 固定フィルタ
固定フィルタは次式により定義される．

1

H 1( f ) =
1–

f0
f

2

– 2h

f0
i
f

(4)

1

⋅

f1
f

1+

2

ここに f0=1/6(Hz), h=0.552，f1=0.1(Hz)である．
このフィルタは，得られる変位波形が気象庁の 1 倍強震計（T=6 秒，h=0.552）の変位波形と比較できるように設定されたものである．フィルタの遮断周波数(3 dB down)は0.154Hz である．なお，ここで言う「フィルタの遮断周波

図-24 式(6)の意味するもの

数(3 dB down)」とは，フィルタの値がフラット部の 10-3/20倍まで低下する周波数のことである．固定フィルタを図の SN 比が良好な記録に対しては小さめの fC が設定され，-21 に示す．

そうでない記録に対しては大きめの fC が設定される．1/T(2) パラメタ付きフィルタ

以下の周波数成分については，fC の決定の際には考慮に入

このフィルタは次式により定義される．

f
H 2( f ) = 1 – exp –fc

れない．
なお，E は以下に示す値である．

2

2

(5)

ERS-F,G 型強震計：
E = ( p×0.001)×0.02236 (Gal)
ここに p は ERS-F,G 型強震計の感度(Gal/215)

このフィルタの遮断周波数は 1.36fC(3 dB down)である．パ

ERS-GV 型強震計：

ラメタ付きフィルタを図-22 および図-23 に示す．
パラメタ
fC は，低周波側での SN 比が良好な記録に対しては小さなE = ( p×0.001)×0.07071 (Gal)値，そうでない記録に対しては大きな値とする．具体的にここに p は ERS-GV 型強震計の感度(Gal/215)

は，以下に述べるσと E が等しくなるように fC を設定する．

SMAC-MDU 型強震計：
E = 2×0.00707(Gal)まず，σは次式により定義される．

σ2 = 1

T

∞

X f

2

1 - exp - fT 2-∞

4

1 - H2 f

2

Datol-100 型強震計：

df

E = 2×0.01257(Gal)(6)

Omni 型強震計：
E = 2×0.002236(Gal)

ここに T はオリジナル加速度波形の長さ，X(f)はオリジナ

Basalt 型強震計：

ル加速度波形のフーリエ変換である．式(6)は，図-24 に示

E = 2×0.00707(Gal)すように，オリジナル加速度波形から計算したフーリエス
ペクトルの自乗を 1/T～fC の範囲で積分した値を示してい2

各強震計の E 値は，非地震時に取得した記録のノイズレる．これが所定の値（E ）となるように fC が設定されるのベルを調べて定めたものである．ERS-F,G 型（フォースフ

で，低周波成分の振幅が大きい記録，すなわち低周波側でィードバック）と ERS-GV 型（速度帰還型）の低周波側の- 15 -

ノイズレベルには差が認められたので，両者の E 値は異なる値としている．
ここで用いる二つのハイパスフィルタはいずれもなだ
らかな周波数特性を持っており，遮断周波数の前後の成分
が時間領域に変換したときに不自然な形をとって現れるのを防ぐことができる．
4.5 解析のアウトプット(1) 加速度，速度，変位各種の加速度波形と速度波形，変位波形が付録 DVD に収められている．先に述べたように，オリジナル加速度波

図-25

1 月 30 日に細島港で観測された加速度波形

形とは基線補正のみを行った波形，補正加速度波形とは，計器特性に関する補正を行った上で，SN 比の十分でないはすべて水平 2 成分のベクトル和をとりバンド幅 0.05Hz

低周波成分をハイパスフィルタにより削除した波形，の Parzen ウインドウを適用したものである．図-26（a）でSMAC-B2 相当加速度波形とは，港湾地域強震観測で長くは，比較のため，最寄りの他機関の観測点である K-NET

用いられてきた SMAC-B2 型強震計による記録と比較可能

日向における同じ地震の記録のフーリエスペクトルを重ねなように周波数成分を調整した波形のことである．速度と書きしている．なお，細島-G と K-NET 日向の位置関係を

変位は，固定フィルタによるものとパラメタ付きフィルタ

図-26（f）に示す．両地点におけるフーリエスペクトルをによるものの双方を提示している．

比較すると，1.2Hz 付近では，細島-G の方がはるかに大き(2) フーリエスペクトルい値を示している．

フーリエスペクトルは補正加速度から FFT により計算さ過去に細島-G と K-NET 日向で観測された記録のフーリ

れる．

エスペクトルも，同様の傾向を示している．図-26（b）－(3) 応答スペクトル（d）は，過去に細島-G と K-NET 日向で観測された主な記

応答スペクトルは補正加速度から計算される．絶対加速録として，2001 年芸予地震（M6.7），2005 年福岡県西方沖

度比，絶対加速度，相対速度，相対変位の 4 種類の応答がの地震（M7.0），2006 年 6 月 12 日大分県北部の地震（M6.2）

0%，1%，5%の 3 種類の 1 自由度系に対して計算される．

の記録のフーリエスペクトルを示したものであるが，いずれの記録においても，1.2Hz 付近では，細島-G の方がはる

5. 2012 年に取得された代表的な記録についてかに大きい値を示している．図-26（e）は，今回の地震を含め，両地点のフーリエスペクトルの比率を示したものである．ただし，今回の地震に対しては，0.4Hz 以下で記録

例年，強震観測年報では，その年に得られた代表的な記録について一章を設けて紹介してきている．本章では，まの精度が十分でないと考えられるため，0.4Hz 以上に対しず，1 月 30 日に日向灘で発生した地震（M4.9）の記録につてプロットを行っている．これを見ると，両地点のスペク

いて紹介し，次に，8 月 25 日に十勝地方南部で発生した地

トル比は地震によらずほぼ同様であるが，今回の地震に対震（M6.1）の記録について紹介する．

してやや大きい値を示している．

5.1

5.2

1 月 30 日日向灘の地震（M4.9）の記録

8 月 25 日十勝地方南部の地震（M6.1）の記録

この地震は 8 月 25 日 23 時 16 分に十勝地方南部の東経

この地震は 1 月 30 日 3 時 18 分に日向灘の東経 132°0.2′，
北緯 32°36.8′，深さ 39km を震源として発生した M4.9

143°6.6′，北緯 42°19.6′，深さ 49km を震源として発の地震であり，宮崎県と大分県で最大震度 4 を記録した．

生した M6.1 の地震であり，十勝支庁と日高支庁で最大震

港湾地域強震観測では細島港と別府港で記録が得られた．

度 5 弱を記録した．港湾地域強震観測では釧路，十勝，浦

このうち細島港の記録は最大加速度が 138.6Gal と比較的大河，苫小牧，室蘭，函館の各港で記録が得られた．このう

きかった．

ち十勝港の記録は最大加速度が 188.9Gal と最も大きかった．

図-25 に細島-G で観測された水平 2 成分の加速度波形を

図-27 に十勝-G で観測された水平 2 成分の加速度波形を示す．これをもとに，フーリエスペクトルを求めたものを示す．これをもとに，フーリエスペクトルを求めたものを

図-26（a）に示す．なお，以下に示すフーリエスペクトル図-28（a）に示す．図-28（a）では，比較のため，最寄り- 16 -

図-26

細島-G と K-NET 日向で観測されたこれまでの主な記録のフーリエスペクトルとそれらの比（下段は両地点の位置関係）- 17 -

図-28

十勝-G と K-NET 広尾で観測されたこれまでの主な記録のフーリエスペクトルとそれらの比（下段は両地点の位置関係）- 18 -

図-27

8 月 25 日に十勝港で観測された加速度波形の他機関の観測点である K-NET 広尾における同じ地震の記録のフーリエスペクトルを重ね書きしている．なお，十勝-G と K-NET 広尾の位置関係を図-28（f）に示す．両地
点におけるフーリエスペクトルを比較すると，5Hz 付近では，十勝-G の方が小さめの値を示している．
過去に十勝-G と K-NET 広尾で観測された記録のフーリ

図-29

浦河港で観測された記録の地表と地中のスペクト

エスペクトルも，同様の傾向を示している．図-28（b）－

ル比．実線は 8 月 25 日の記録．破線は振幅の小さ（d）は，過去に十勝-G と K-NET 広尾で観測された主な記

い記録．

録として，1999 年 5 月 13 日釧路地方の地震（M6.3），2003
年十勝沖地震（M8.0），十勝沖地震の最大余震（M7.1）のことができるものと考えられる．

記録のフーリエスペクトルを示したものであるが，いずれの記録においても，5Hz 付近では，十勝-G の方が小さめの

6. まとめ値を示している．図-28（e）は，今回の地震を含め，両地点のフーリエスペクトルの比率を示したものである．これ

本資料は，1962 年より実施されている港湾地域強震観測を見ると，両地点のスペクトル比は地震によらずほぼ同様の記録のうち，2012 年に得られた記録について報告するもである．

のである．得られた記録は地点毎に分類され，地震諸元，

最後に，この地震の際に浦河港で観測された表層地盤の記録番号，最大加速度等とともに付録 DVD の強震観測表

非線形挙動について述べる．この地震による浦河港の記録にまとめられている．強震観測表の地震諸元は気象庁の地に対して，地表と地中のフーリエスペクトル（水平 2 成分

震・火山月報（カタログ編）に基づいている．強震観測表のベクトル和をとりバンド幅 0.05Hz のパーセンウインドにまとめられている記録の中で，最大加速度が 20Gal を越ウを適用したもの）の比を求めたものが図-29 の実線であえる記録については加速度波形を示している．また，最大る．それに対し，図-29 の破線は，地盤が線形の範囲で挙加速度が 50Gal を越える比較的大きな記録については，加動していると考えられる振幅の小さい記録に対して，同じ速度波形に加え，速度波形，変位波形，フーリエスペクト

ように浦河港の地表と地中でのフーリエスペクトルの比をル，応答スペクトルも併せて示している．

プロットしたものである．これを見ると，8 月 25 日の記録

（2013年11月8日受付）

では，スペクトル比のピーク周波数が，線形時の 5.1Hz から 4.5Hz まで低下している．従って，8 月 25 日の地震では，謝辞

浦河港の強震観測地点では地盤に非線形挙動が生じたと考

港湾地域強震観測は，当所の他，国土交通省港湾局，東

えられ，表層地盤（正確には地表と地中の強震計で挟まれ北地方整備局，関東地方整備局，北陸地方整備局，中部地

た範囲）の S 波速度は約 12%低下したと考えられる．
また，

方整備局，近畿地方整備局，中国地方整備局，四国地方整地盤のせん断剛性は S 波速度の自乗に比例するので，地盤

備局，九州地方整備局，北海道開発局，内閣府沖縄総合事のせん断剛性は約 22%低下したと考えられる．これらの情務局，東京都港湾局，静岡県港湾総室，大阪市港湾局，神報は今後，地盤の地震応答計算プログラムの検証に用いる

戸市みなと総局および宮崎県港湾課の各機関の協力の下に- 19 -

16）倉田栄一・石坂徳三・土田

実施されています．地震動研究チームの林公美さん，宮田

肇：港湾地域強震観測年

報（1974），港湾技研資料 No.202，1975 年 3 月．

さよ子さんには記録の処理などで大きく貢献していただい17）倉田栄一・井合ています．第 5 章では比較のため防災科学技術研究所の進・土田肇：港湾地域強震観測年

報（1975），港湾技研資料 No.236，1976 年 3 月．

K-NET の記録を利用しました．記して謝意を表します．

18）倉田栄一・井合進・土田

肇：港湾地域強震観測年

参考文献

報，補遺（1963~1975，上下動成分），港湾技研資料 No.250，

1）菅野高弘・三籐正明・及川研：兵庫県南部地震による港

1976 年 12 月．

湾施設の被害考察（その 8）ケーソン式岸壁の被災に関

19）倉田栄一・井合20）倉田栄一・井合pp.207-252．

21）倉田栄一・井合第 36 巻，第 2 号，1997 年 6 月，pp.41-86．

22）倉田栄一・野田節男：港湾地域強震観測年報（1981），

湾空港技術研究所資料 No.1112, 2005 年 12 月．

港湾技研資料 No.426，1982 年 6 月．

4）野津厚・菅野高弘：スペクトルインバージョンに基づく

23）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

南西諸島の強震観測地点におけるサイト増幅特性，港湾
空港技術研究所資料 No.1149, 2007 年 3 月．

報（1982），港湾技研資料 No.446，1983 年 6 月．

5）野津厚・菅野高弘：スペクトルインバージョンに基づく

24）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年
報（1983），港湾技研資料 No.487，1984 年 6 月．

道北の強震観測地点におけるサイト増幅特性，港湾空港
技術研究所資料 No.1214, 2010 年 6 月．

25）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

6）野津厚・盛川仁：表層地盤の多重非線形効果を考慮した

報（1984），港湾技研資料 No.519，1985 年 6 月．
26）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

pp.361-374．

報（1985），港湾技研資料 No.547，1986 年 6 月．

肇・山田逓一郎・倉田栄一・須藤克子：港湾地域

27）倉田栄一・井合

強震観測年報（1963･1964），港湾技研資料 No.55，1968
年 9 月．

報（1986），港湾技研資料 No.588，1987 年 6 月．
報（1987），港湾技研資料 No.618，1988 年 6 月．

肇・山田逓一郎・倉田栄一・須藤克子：港湾地域

29）倉田栄一・井合肇・倉田栄一・須藤克子：港湾地域強震観測年報

30）倉田栄一・井合肇・倉田栄一・須藤克子：港湾地域強震観測年

31）倉田栄一・井合報（1968），港湾技研資料 No.98，1970 年 3 月．
肇・倉田栄一・須藤克子：港湾地域強震観測年

32）倉田栄一・井合33）松永康男・桜井博孝・森田年一・井合
年 6 月．

34）宮田正史・佐藤幸博・一井康二・森田年一・井合

肇：港湾地域強震観測年

進：

港湾地域強震観測年報（1994），港湾技研資料 No.840，

肇：港湾地域強震観測年

報（1972），港湾技研資料 No.160，1973 年 3 月．
15）倉田栄一・石坂徳三・土田進：港湾地域

強震観測年報（1992&1993），港湾技研資料 No.776，1994

肇：港湾地域強震観測年

報（1971），港湾技研資料 No.136，1972 年 3 月．
14）倉田栄一・石坂徳三・土田進：港湾地域強震観測年報（1991），

港湾技研資料 No.727，1992 年 6 月．

報（1970），港湾技研資料 No.116，1971 年 3 月．
13）倉田栄一・石坂徳三・土田進：港湾地域強震観測年報（1990），

港湾技研資料 No.705，1991 年 6 月．

報（1969），港湾技研資料 No.100，1970 年 6 月．
肇・倉田栄一・須藤克子：港湾地域強震観測年

進：港湾地域強震観測年報（1989），

港湾技研資料 No.676，1990 年 6 月．

（1967），港湾技研資料 No.64，1969 年 3 月．

12）土田進：港湾地域強震観測年報（1988），

港湾技研資料 No.649，1989 年 6 月．

年 12 月．

11）土田進・野田節男：港湾地域強震観測年

28）倉田栄一・野田節男・樋口豊志：港湾地域強震観測年

強震観測年報（1965･1966），港湾技研資料 No.62，1968

10）土田進・横山淑子・野田節男：港湾地域

月．

国の港湾等の強震観測地点におけるサイト増幅特性，港9）土田

肇：港湾地域

強震観測年報（1980），港湾技研資料 No.374，1981 年 63）野津厚・長尾毅：スペクトルインバージョンに基づく全

8）土田進・横山淑子・土田

年 6 月．

ケーソン式岸壁の挙動の有効応力解析，港湾技研報告，

7）土田

肇：港湾地域強震観測年

強震観測年報（1978･1979），港湾技研資料 No.338，1980

2）一井康二・井合進・森田年一：兵庫県南部地震における経験的グリーン関数法，地震2，第55巻，2003年3月，

進・土田報（1976･1977），港湾技研資料 No.287，1978 年 3 月．

する模型振動実験，港湾技研資料，No.813，1995 年 9 月，

1996 年 6 月．
35）佐藤幸博・一井康二・井合報（1973），港湾技研資料 No.181，1974 年 3 月．

進・星野裕子・佐藤陽子・

宮田正史・森田年一：港湾地域強震観測年報（1995&1996），- 20 -

震の港湾地域における強震記録，港湾技研資料 No.442，

港湾技研資料 No.909，1998 年 9 月．
36）佐藤幸博・一井康二・井合1983 年 3 月．

進・星野裕子・佐藤陽子：54）倉田栄一・福原哲夫・野田節男：昭和 58 年（1983 年）

港湾地域強震観測年報（1997），港湾技研資料 No.936，
1999 年 6 月．

日本海中部地震の港湾地域における強震記録，港湾技研

37）佐藤幸博・一井康二・星野裕子・佐藤陽子・井合長尾

資料 No.458，1983 年 9 月．

進・55）倉田栄一・福原哲夫・野田節男：昭和 59 年（1984 年）

毅：港湾地域強震観測年報（1998），港湾技研資8月7日

料 No.942，1999 年 9 月．

湾技研資料 No.503，1984 年 12 月．

38）野津厚・深澤清尊・佐藤陽子・玉井伸昌・菅野高弘：56）倉田栄一・野田節男・樋口豊志：昭和 62 年（1987 年）

港湾地域強震観測年報（1999&2000），港湾空港技術研究

12 月 17 日

所資料 No.1016，2002 年 3 月．
震観測年報（2001），港湾空港技術研究所資料 No.1019，

57）松永康男・桜井博孝・森田年一・井合2002 年 6 月．

No.777，1994 年 6 月．
58）松永康男・桜井博孝・森田年一・井合

港湾地域強震観測年報（2002），港湾空港技術研究所資料 No.1054，2003 年 9 月．

資料 No.778，1994 年 6 月．

震観測年報（2003），港湾空港技術研究所資料 No.1084，59）佐藤幸博・宮田正史・一井康二・森田年一・井合2004 年 9 月．

進：1994 年北海道東方沖地震の港湾地域における強震記録，

42）野津厚・菅野高弘：港湾地域強震観測年報（2004），港湾空港技術研究所資料 No.1109，2005 年 9 月．

港湾技研資料 No.853，1996 年 12 月．
60）佐藤幸博・一井康二・宮田正史・森田年一・井合
港湾技研資料 No.892，1997 年 12 月．

湾空港技術研究所資料 No.1136，2006 年 9 月．
44）野津厚・菅野高弘：港湾地域強震観測年報（2006），港

61）佐藤幸博・一井康二・星野裕子・佐藤陽子・宮田正史・

湾空港技術研究所資料 No.1164，2007 年 9 月．

森田年一・井合62）土田

63）倉田栄一・土田64）倉田栄一・石坂徳三：港湾地域強震観測地点資料（その 3），港湾技研資料 No.156，1973 年 3 月．

淳：港湾地域強震観測年報（2010），港湾空港技術研究所資料 No.1243，2011 年 12 月．

65）横山淑子・倉田栄一：港湾地域強震観測地点資料（その 4），港湾技研資料 No.298，1978 年 6 月．

淳：港湾地域強震観測年報（2011），港

湾空港技術研究所資料 No.1266，2013 年 3 月．

66）横山淑子・倉田栄一：港湾地域強震観測地点資料（その 5），港湾技研資料 No.351，1980 年 9 月．

肇・倉田栄一・須藤克子：1968 年十勝沖地震とその余震の港湾地域における強震記録，港湾技研資料67）一井康二・佐藤幸博・佐藤陽子・星野裕子・井合No.80，1969 年 6 月．
51）倉田栄一・井合

進・土田

進：

港湾地域強震観測地点資料（その 6），港湾技研資料No.935，1999 年 6 月．

肇：1978 年伊豆大島近海の地震の港 湾地域における強震記録， 港湾技研資料68）木下繁夫：サーボ型地震計，地震2，第50巻，1998年3

No.317，1979 年 3 月．
52）倉田栄一・井合肇・須藤克子：港湾地域強震観測地点資料（その 2），港湾技研資料 No.107，1970 年 12 月．

淳：港湾地域強震観測年報（2009），港湾空港技術研究所資料 No.1223，2010 年 12 月．

肇・山田逓一郎・倉田栄一：港湾地域強震観測

地点資料（その 1），港湾技研資料 No.34，1967 年 11 月．

46）野津厚・菅野高弘：港湾地域強震観測年報（2008），港湾空港技術研究所資料 No.1207，2010 年 3 月．

進：1995 年兵庫県南部地震の港湾地域における強震記録，港湾技研資料 No.907，1998 年 6 月．

45）野津厚・菅野高弘：港湾地域強震観測年報（2007），港湾空港技術研究所資料 No.1184，2008 年 9 月．

進：1994 年三陸はるか沖地震の港湾地域における強震記録，

43）野津厚・菅野高弘：港湾地域強震観測年報（2005），港50）土田進：1993 年北海道南西沖地震の港湾地域における強震記録，港湾技研

41）野津厚・鈴木嘉秀・早田泰子・菅野高弘：港湾地域強49）野津厚・若井

進：1993 年釧路沖地震の港湾地域における強震記録，港湾技研資料40）野津厚・佐藤陽子・深澤清尊・佐藤泰子・菅野高弘：48）野津厚・若井

千葉県東方沖地震の港湾地域における強震

記録，港湾技研資料 No.619，1988 年 6 月．

39）深澤清尊・佐藤陽子・野津厚・菅野高弘：港湾地域強47）野津厚・若井

日向灘地震の港湾地域における強震記録，港進・横山淑子・土田月，pp.471-483．
肇：1978 年宮

城県沖地震の港湾地域における強震記録，港湾技研資料

69）気象庁：地震・火山月報（防災編），2012 年．
70）気象庁：地震・火山月報（カタログ編），平成 24 年 12No.319，1979 年 6 月．
53）倉田栄一・野田節男：昭和 57 年（1982 年）浦河沖地

月，2012 年．
71）井合- 21 -

進・倉田栄一・土田肇：強震記録の数字化と補正，港湾技研資料 No.286，1978 年 3 月．
72）Susumu Iai and Eiichi Kurata: Integration of strong-motion
accelerograms， Proceedings of the 5th Japan Earthquake
Engineering Symposium，1978，pp.225-232.
73）高橋重雄・戸田和彦・菊池喜昭・菅野高弘・栗山善昭・
山﨑浩之・長尾毅・下迫健一郎・根木貴史・菅野甚活・

富田孝史・河合弘泰・中川康之・野津厚・岡本修・鈴木
高二朗・森川嘉之・有川太郎・岩波光保・水谷崇亮・小濱英司・山路徹・熊谷兼太郎・辰巳大介・鷲崎誠・泉山
拓也・関克己・廉慶善・竹信正寛・加島寛章・伴野雅之・
福永勇介・作中淳一郎・渡邉祐二：2011 年東日本大震災
による港湾・海岸・空港の地震・津波被害に関する調査
速報，港湾空港技術研究所資料 No.1231，2011 年 4 月．
74）Kinoshita, S. : Kyoshin Net (K-net), Seim. Res. Lett., Vol.69,
1998, pp.309-332.
75）Iai, S., T. Morita, T. Kameoka, Y. Matsunaga and K. Abiko:
Response of a dense sand deposit during 1993 Kushiro-oki
earthquake, Soils and Foundations, Vol.35, 1995, pp.115-131.
76）野津厚・井合進・一井康二・沼田淳紀：ケーソン式岸壁の変形に寄与する地震動の周波数成分，レベル 2 地震動に対する土構造物の耐震設計シンポジウムおよび講習会テキスト，2000 年 8 月，pp.311-318．
77）野津厚・若井

淳：東日本大震災で被災した港湾における地震動特性，港湾空港技術研究所資料 No.1244, 2011年 12 月．

- 22 -

強震観測担当者（2012年１月～12月）
東北地方整備局
青森港湾事務所

玉山

具夫

斉藤

雅志

八戸港湾・空港整備事務所

滝沢

洋一

齊藤

聡洋

福田

良介

山田

裕之

佐藤

和敏
原茂

雅光

釜石港湾事務所
釜石港湾事務所

宮古港事務所

塩釜港湾・空港整備事務所

外久保

小名浜港湾事務所

根本

俊彦

佐藤

匡

秋田港湾事務所

菅原

将徳

酒田港湾事務所

百澤

康雄

渡辺

義

小名浜港湾事務所

相馬港出張所裕一

瀬川哲

北陸地方整備局
新潟港湾・空港整備事務所
伏木富山港湾事務所

新湊出張所

波多野

宮崎

賢治

佐藤

金沢港湾・空港整備事務所

杉原

友和

南

敦賀港湾事務所

波多野

雅也

雅也
由浩

靖彦

岡本

博

秀治

関東地方整備局
鹿島港湾・空港整備事務所

笹森

秀一郎

笹森

秀一郎

千葉港湾事務所

石岡勉

四戸

横浜港湾空港技術調査事務所

佐藤

一央

堀

京浜港湾事務所

中村

弘樹

吉田

明俊

鹿島港湾・空港整備事務所

常陸那珂港事務所

貞治

中部地方整備局
清水港湾事務所

本田宗隆

清水港湾事務所

下田港事務所對木努

清水港湾事務所

御前崎港事務所

冨本

正

前田

千

横井

良典

梅田

至人

野上

周嗣

名古屋港湾事務所

澤田

隆志

牛場

茂友

名古屋港湾空港技術調査事務所

吉村藤謙

四日市港湾事務所

船橋香

和歌山港湾事務所

竹村知子

神戸港湾事務所

渡慶次

藤井

慶一朗

尾崎

靖

三河港湾事務所
三河港湾事務所

衣浦港事務所

近畿地方整備局
賀則

中国地方整備局
境港湾・空港整備事務所

仲濱弘平

広島港湾・空港整備事務所

石橋

伸司

23

五十嵐
村上

渉
栄喜

四国地方整備局
松山港湾・空港整備事務所

泉保佑介

小松島港湾・空港整備事務所

斉藤

祐一

前田

千

高知港湾・空港整備事務所

真砂

澄男

西本

朋弘

山本

龍三

山本

弘幸

松尾

博文

畠中

丈

小野

章

竹口

はや人

高坂

誠

宮崎港湾・空港整備事務所

森満

賢治

小池

二三勝

志布志港湾事務所

野田巧

鹿児島港湾・空港整備事務所

荻

千葉

政幸

安永

健太

九州地方整備局
博多港湾・空港整備事務所
別府港湾・空港整備事務所

日巻

繁

添田

宏

三浦

豪

定治

北海道開発局
釧路開発建設部釧路港湾事務所

山内弘明

釧路開発建設部根室港湾事務所菊地聡一

帯広開発建設部築港対策官付鈴木孝信

室蘭開発建設部

浦河港湾事務所

山岡

広幸

室蘭開発建設部

苫小牧港湾事務所

鈴木貴秀

室蘭開発建設部室蘭港湾事務所

大沢

佳宏

伊藤

雅和

小樽開発建設部小樽港湾事務所

杉山

盛行

貞島

一雄

佐伯

茂

三浦

正人

函館開発建設部函館港湾事務所

明石

卓弥

佐藤

亘

函館開発建設部

江差港湾事務所

藤由

幸大

本山

賢司

新岡

博文

留萌開発建設部留萌港湾事務所

佐々木

上村

剛志

岡本

健太郎

網走開発建設部

網走港湾事務所

賀川智章

網走開発建設部

紋別港湾事務所

鈴木

慶律

高橋

博

中岡

泰典
憲

當銘

須賀子

智之

沖縄総合事務局
那覇港湾・空港整備事務所
那覇港湾・空港整備事務所

中城湾港出張所

伊良波

平良港湾事務所

城間

由樹

野瀬

晴生

石垣港湾事務所

田港

朝之

山田

文人

都道府県
東京都港湾局森

大阪市港湾局

黒田

剛

本多

誠司

神戸市みなと総局

能勢

正義

藤井

研一

成本

克彦

静岡県田子の浦港管理事務所

佐野光政

宮崎県土木部北部港湾事務所

黒木

成一

児玉

勝美

興梠

真和

24

洋

港湾空港技術研究所資料

No.1283

２０１４．３

編集兼発行人

独立行政法人港湾空港技術研究所

発

行

所

独立行政法人港湾空港技術研究所
横 須 賀 市 長 瀬 ３ 丁 目 １ 番 １ 号
TEL. 046(844)5040 URL. http://www.pari.go.jp/

印

刷

所

株 式 会 社 シ ー ケ ン

C （2014）by PARI
Copyright ○All rights reserved. No part of this book must be reproduced by any means without the written
permission of the President of PARI
この資料は、港湾空港技術研究所理事長の承認を得て刊行したものである。したがって、本報告書の全部または一部の転載、複写は港湾空港技術研究所理事長の文書による承認を得ずしてこれを行ってはならない。