What could be easier?

Power Monitor OPM150

Highlights:
Low cost
Easy to use
Powerful
Our offer in Detail:
The OPM150 is a versatile power monitor for use in the lab and for laser servicing as well as for OEM applications.
Heads are available covering 190-1100nm, 800-1550nm,
800-1650nm and 400-1600nm. Each head is automatically recognized and its individual calibration data areuploaded to the system. In addition, the user may load up to 5 separate filter curves. Thus the dynamic range andfunctionality of the unit can be expanded while ensuring proper calibration.
The graphical user interface is intuitive to use and easy to read! Thesoftware includes a scope function, data logging and a large, digital
display – perfect for daily use in the lab or in the field. Further
functions such as autogain and various storage formats
are also included.
The OPM150 is small enough to fit into your pocket
and is USB powered and controlled. No extra power
supply required, no extra cables on the table. Not only
the small size qualifies this instrument for OEM
applications. It is very simple and flexible to integrate into your project. The unit comes delivered with drivers for
direct communication or to be used as a virtual COM-port. Furthermore, we provide a full software development kit
including the source code for the GUI application as well as a demo LabView-VI.

Specifications• USB powered and controlled• 30 Hz update rate with GUI, 1000 Hz as data logger• 5 gain ranges from 1µW to 10mW full scale (<1nW resolution!)• Plenty of accessories to enhance the ease of application

Your problem is our challenge – flexibility is our standard:
We will gladly adapt, for example, the wavelength or the aperture to suit your application.
Let us know your requirements.
1

OPM150_100209_Engl.doc

Base module including one UV-Silicon detector head, software and carrying case.
LabView is a registered trade mark of National Instruments Corporation.

Artifex Engineering e.K.
Dortmunder Str. 16-18
26723 Emden, Germany

General Manager: Dr. Steven Wright
Registry number: HRA 200036

Tel: +49-(0)4921-58908-0
eMail: info@afx-eng.com
Web: http://www.afx-eng.com/

Ordering Information
Base module: OPM150
Detector heads:
Full order code:

Diode material (d):Options

Description

UVSG5G10IGA
VIGA
VIGAFC

190-1100nm; 10x10mm
800-1550nm; φ 5mm
800-1550nm; φ 10mm
800-1650nm; φ 3mm
400-1600nm; φ 1mm
400-1600nm; FC receptacle

OPM

150dSpecifications
SiPARAMETER
INPUT

CONDITIONS

Min

Wavelength range

UVS Detector

190

Max

UNITS

1100nm252500250252500mWmW
µW
µW
µW
µW
nW
nW
nW
pW

2

Power ranges (full scale)

Noise equivalent power (NEPRMS)

Typ

3.0
1100
101
Min @ 900 nm
Max @ 190 nm

5500
505500OUTPUT
Connectors

DB9 and USB

Bandwidth

10kHz

GUI-Control

30S/s

Direct polling

1000

Sampling rate
Accuracy± 10%

Reproducibility±3%Linearity± 0.2± 0.4dB

SUPPLYTypeUSB

DIMENSIONS
60 x 81 x 36 mm (B x L x H)

Accessories:

2mm- calibrated filters- adapters for standard optomechanic systems- reducing apertures- fibre receptacles- calibrated beamsplitter (coming soon)

Linearity guaranteed to 1.5mW (1.8dBm). For higher powers, use our calibrated OD filters or a detector type G5 or G10.

OD1 up to 15mW
OD2 up to 150mW
OD3 up to 1.5W