SENSOR TECHNOLOGIESDIGITAL COMPONENT SENSORDEVELOPMENT TOOLSRATE AND INERTIAL SENSORSPAGE 49

PAGE 4

FLOW SENSORSSCANNERS AND
SYSTEMSPAGE 8

PAGE 51

FORCE SENSORSSPEED SENSORSPAGE 10

PAGE 53

HUMIDITY SENSORSTEMPERATURE SENSORSPAGE 16

PAGE 59

LIQUID LEVEL SENSORSTORQUE SENSORSPAGE 20

PAGE 67

PHOTO OPTIC SENSORSULTRASONIC SENSORSPAGE 22

PAGE 69

PIEZO FILM SENSORSVIBRATION SENSORSPAGE 24

PAGE 71

POSITION SENSORSWATER LEVEL SENSORSPAGE 26

PAGE 79

PRESSURE SENSORSPAGE 40

SS-TS-TE101
06/2019te.com/sensors

PAGE 3

DIGITAL COMPONENT SENSOR DEVELOPMENT TOOLSDIGITALCOMPONENTSENSORDEVELOPMENTTOOLSWIRELESS DEMO AND DEVELOPMENT KITS

Environmental Sensor Tag

M5600

U5600

Type

Humidity, Temperature, Pressure

Pressure

Pressure

Specifications• 0 - 100% RH• 300 to 1,200 mbar

• 50 - 15K psi• Type G/S/C• 2 - 10K psi• Type G/S/C/A

Communication
Interface

Standard 2.4 GHz wireless communication

Standard 2.4 GHz wireless communication

Standard 2.4 GHz wireless communication

Application

iOS 7.0+ Android™ 4.3+

iOS 7.0+ Android™ 4.3+

iOS 7.0+ Android™ 4.3+

Operating Temp.

20°C to 85°C

-20°C to 85 °C-20°C to 85°C

PAGE 4
DIGITAL COMPONENT SENSORDEVELOPMENT TOOLS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

DIGITAL COMPONENT SENSOR DEVELOPMENT TOOLSSHIELDS AND HATS

Arduino Weather Shield

Raspberry PiTM Sensors Weather Hat

PicTail PlusType

Humidity, Temperature, Pressure

Humidity, Temperature, Pressure

Humidity, Temperature, Pressure

Sensors

HTU21D, MS5637, MS8607, TSYS01*, TSD305

HTU21D, MS5637, TSYS01*, TSD305

HTU21DF, MS5637, TSYS01*, MS8607

Specifications• 0 - 100% RH• 20°C to 85°C
• 300 to 1,200 mbar

• 0 - 100% RH• 20°C to 85°C
• 300 to 1,200 mbar

• 0 - 100% RH• -20°C to 85°C
• 300 to 1,200 mbar

Communication
Interface

I²C

I²C

I²C

Partner Board

Arduino / Genuino

Raspberry PiTM

Microchip Explorer 16

WING BOARDSHTU21DMS5637MS8607

TSYS01*

TSYS02D*

KMA36(A)Type

Humidity

Pressure

Pressure, Temperature,
Humidity

Temperature

Temperature

Angular Position

Specifications• 0 to 100% RH• -40°C to 125°C• 3.3 to 5.5 V• 10 to 2,000 mbar
• -40°C to 85°C
• 1.5 to 3.6 V• 10 to 2,000 mbar
• -40°C to 85°C
• 0 to 100% RH• 1.5 to 3.6 V• -40°C to 125°C• 2.2 to 3.6 V• -40°C to 125°C• 1.5 to 3.6 V• 0 to 360°
• -25°C to 85°C
• 2.9 to 6.0 V

Accuracy±3% RH±2 mbar±3% RH, ±2
mbar, ±1.0°C±0.1°C±0.2°C±0.1°

Communication
Interface

I²C

I²C

I²C

I²C

I²C

I²C

Compatibility

Configured to operatewith the Xplained Pro
development platform

Configured to operatewith the Xplained Pro
development platform

Configured to operatewith the Xplained Pro
development platform

Configured to operatewith the Xplained Pro
development platform

Configured to operatewith the Xplained Pro
development platform

Configured to operatewith the Xplained Pro
development platform*Temperature System Sensor (TSYS)

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 5
DIGITAL COMPONENT SENSORDEVELOPMENT TOOLSDIGITAL COMPONENT SENSOR DEVELOPMENT TOOLSPERIPHERAL MODULES
Digilent Pmod™

HTU21DMS5637MS8607

TSYS01*

TSYS02D*

KMA36(A)Type

Humidity

Pressure

Pressure, Temperature,
Humidity

Temperature

Temperature

Angular Position

Specifications• 0 to 100% RH• -40°C to 125°C• 3.3 to 5.5 V• 10 to 2,000 mbar
• -40°C to 85°C
• 1.5 to 3.6 V• 10 to 2,000 mbar
• -40°C to 85°C
• 0 to 100% RH• 1.5 to 3.6 V• -40°C to 125°C• 2.2 to 3.6 V• -40°C to 125°C• 1.5 to 3.6 V• 0° to 360°
• -25°C to 85°C
• 2.9 to 6.0 V

Accuracy±3% RH±2 mbar±3% RH, ±2 mbar,±1.0°C±0.1°C±0.2°C±0.1°

Communication
Interface

I²C

I²C

I²C

I²C

I²C

I²C

Compatibility

Development
systems compatible
with Digilent Pmod™
connections

Development
systems compatible
with Digilent Pmod™
connections

Development
systems compatible
with Digilent Pmod™
connections

Development
systems compatible
with Digilent Pmod™
connections

Development
systems compatible
with Digilent Pmod™
connections

Development
systems compatible
with Digilent Pmod™
connections

MS5611

MS5837

MS5805

TSD305Type

Pressure

Pressure

Pressure

TemperatureSpecifications• 10 to 1,200 mbar
• -40°C to 85°C
• 1.5 to 3.6 V• 10 to 2,000 mbar
• -40°C to 85°C
• 1.5 to 3.6 V• 10 to 2,000 mbar
• -40°C to 85°C
• 1.8 to 3.6 V• -10°C to +85°C• 1.68 to 3.6 V

Accuracy±2 mbar±2 mbar±2 mbar±1°C

Communication
Interface

I²C

I²C

I²C

I²C

Compatibility

Development systems compatible
with Digilent Pmod™ connections

Development systems compatible
with Digilent Pmod™ connections

Development systems compatible
with Digilent Pmod™ connections

Development systems compatible
with Digilent Pmod™ connections

PAGE 6
DIGITAL COMPONENT SENSORDEVELOPMENT TOOLS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

DIGITAL COMPONENT SENSOR DEVELOPMENT TOOLSGROVE SYSTEM

KMA36

TSYS01*

TSYS02*

MS5637MS8607Type

Angular Position

Temperature

Temperature

Pressure

Pressure, Temperature,
Humidity

Specifications• 0 to 360°
• -25°C to 85°C
• 5.0 V• -40°C to 125°C• 5.0 V• -40°C to 125°C• 5.0 V• 10 to 2,000 mbar
• -40°C to 85°C
• 5.0 V• 10 to 2,000 mbar
• -40°C to 85°C
• 0 to 100% RH• 5.0 V

Accuracy±0.1°±0.1°C±0.2°C±2 mbar±3% RH, ±2 mbar, ±1.0°C

Comm. Interface

I²C

I²C

I²C

I²C

I²C

Compatibility

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

HTU21DMS5611

MS5837

MS5805

TSD305Type

Humidity

Pressure

Pressure

Pressure

TemperatureSpecifications• 0 to 100% RH• -40°C to 125°C• 5.0 V• 10 to 1,200 mbar
• -40°C to 85°C
• 5.0 V• 10 to 2,000 mbar
• -40°C to 85°C
• 5.0 V• 10 to 2,000 mbar
• -40°C to 85°C
• 5.0 V• -10°C to +85°C• 5.0 V

Accuracy±3% RH±2 mbar±2 mbar±2 mbar±1°C

Communication
Interface

I²C

I²C

I²C

I²C

I²C

Compatibility

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

Development platform
compatible withgrove systems

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 7
DIGITAL COMPONENT SENSORDEVELOPMENT TOOLSFLOW SENSORSFLOWSENSORS

PAGE 8
FLOW SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

FLOW SENSORSMASS AIR FLOW SENSORSLMM-H03

LMM-H04

Package

HybridHybridType• Hot film anemometer component• Bidirectional• Hot film anemometer component• Unidirectional

Operating Temp.

-40°C to 125°C-40°C to 125°C

Unique Features

High sensitivity at low heater temperatures,
fast response time, true air temperature sensor

High sensitivity at low heater temperatures,
fast response time, true air temperature sensor

Calibration/Accuracy

Dependent on electronics

Dependent on electronics

Dimensions (mm)23 x 10.15 x 1.1

24 x 10.15 x 1.1

Typical Applications

Air intake of combustion engine, spirometer, industrial gas flow

Air intake of combustion engine, spirometer, industrial gas flow

FLOW SWITCHES

FS-01

FS-02

FS-05

FS-06

FS-90/1

Package

Noryl®

Noryl®

Brass

Brass

CopperType

Flow switch for directionof liquid and gas flow

Flow switch for directionof liquid and gas flow

Flow switch for directionof liquid and gas flow

Flow switch for directionof liquid and gas flow

Flow switch for directionof liquid and gas flow

Max. Pressure

10 bar at 20°C

10 bar at 20°C

10 bar at 20°C

10 bar at 20°C

10 bar at 20°C

Operating Temp.

-30°C to 85°C

-30°C to 85°C

-30°C to 100°C-30°C to 100°C-30°C to 85°C

Unique Features

Triac, normally open,
close on flow

SPST reed switch,normally open,
close on flow

Triac, normally open,
close on flow

SPST reed switch,normally open,
close on flow

SPST reed switch,normally open,
close on flow

Dimensions (mm)106 x 32 x 32

106 x 32 x 32

113 x 53 x 36

113 x 53 x 36

153 x 25 x 15

Typical Applications

Mains water control,
power shower,
central heating
systems, circulation
pump protection,
cooling systems

Mains water control,
power shower,
central heating
systems, circulation
pump protection,
cooling systems

Mains water control,
power shower,
central heating
systems, circulation
pump protection,
cooling systems

Mains water control,
power shower,
central heating
systems, circulation
pump protection,
cooling systems

Leak detection, flow
sensing, mains water
control, coolingsystems, circulation
pump protection

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 9FLOW SENSORSFORCE SENSORSFORCESENSORS

PAGE 10FORCE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

FORCE SENSORSLOAD CELLS
Low Cost OEM

FX19

FX29

FS19

PackageLow profile “coin cell” design

Welded miniature compression
load cell, Analog and Digital options

Stainless steel housing with flexible PCB

Operating Mode

Compression

Compression

Compression

Unique Features• Low cost, low strain design• Essentially unlimited cycle life

• Best price to performance• Designed for unlimited cycles
and high over-range• Low cost• Small size and light weightFS Ranges

10 to 200 lbf
50 to 100 Newton

10 to 100 lbf
50 to 500 Newton

500 to 3,000 grams-force
5 to 30 Newton

Max. Over-range

2.5X FS

2.5X FS

Output/Span

100 mV

100 mV, 0.5-4.5 VDC, Digital (I C)

100 mV

Combined Linearity & Hysteresis±1.0% FSO

±1.0% FSO

±1.0% FSO

Operating Temperature

0°C to 50°C

0°C to 50°C

0°C to 40°C

Dimensions (mm)Ø25.00 x 8.00

Ø19.70 x 4.95

Ø9.5 x 3.45

Typical Applications

Consumer OEM, exercise machines,
physical therapy, vending machines,
appliances, pumps, medical devices

Consumer OEM, exercise machines,
physical therapy, vending machines,
appliances, pumps, medical devices

Infusion pump, load sensing, contact
sensing, weighing, household appliances

2X FS
2

FS20

FC22

FC23

Package

Miniature, drop in replacement forindustry standard

Plastic housing, button, flange mounting

Stainless steel housing button
shape for higher weight loads

Operating Mode

Compression

Compression

Compression

Unique Features• Load cell design operates at very
low strains• Not subject to lead die fatigue• Low cost button shape• Essentially unlimited cycle life

• Industry standard low profile
all stainless steel design• Resistant to off-axis loads

FS Ranges

500 to 5,000 grams-force
5 to 50 Newton

10 to 100 lbf
50 to 500 Newton

50 to 2,000 lbf
250 to 1,000 Newton

Max. Over-range

2.5X FS

2.5X FS

2.5X FS

Output/Span

0.5-4.5 VDC, 1.0-4.0 VDC

100 mV, 0.5-4.5 VDC

100 mV, 0.5-4.5 VDC

Combined Linearity & Hysteresis±1.0% FSO

±1.0% FSO

±1.0% FSO

Operating Temperature-40°C to 85°C

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)30.708 x 17.272 x 8.255

Ø26.00 x 42.00 x 19.50

Ø31.75 x 10.20

Typical Applications

Infusion pumps, contact sensing,
medical devices, consumer appliances

Infusion pumps, robotics end-effectors,
exercise machines, contact sensing,appliances

Batch weighing, robotics, assembly line
force, printing presses, pumps, winch
and hoist

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 11
FORCE SENSORSFORCE SENSORSLOAD CELLSStandard

FMT

FN1010

FN2420FN3000, FN3050

FN3002

Package

WasherLoad pin design

Very high capacity
load button

Pan-cake

Very high capacity
dual studOperating Mode

Compression

Tension and compression

Compression

Tension and compression

Tension and compression

Unique Features• High stiffness• Clamping and bolt forces• High temperature option• Keyed anti-rotation slot
• Bidirectional available• Optional watertightconstruction• High stiffness• Optional load button• Optional high level
output module• High stability• All FN3050 havesame housing• Optional high level output• Threaded male fitting• Integrated amplifier• Optional rod end

FS Ranges

4K to 64K lbf
20K to 320K Newton

2K to 400K lbf
10K to 2,000K Newton

4K to 1000K lbf
20K to 5,000K Newton

20 to 200K lbf
100 to 1,000K Newton

2K to 400K lbf
10K to 2,000K Newton

Max. Over-range

1.5X FS

1.5X FS

1.5X FS

1.5X FS (10X FS with stops)

1.5X FS

Output/Span±20 mV±20 mV, 0.5-4.5
VDC, 4-20 mA±20 mV, 0.5-4.5 VDC±20 mV, 0.5-4.5 VDC±20 mV, 0.5-4.5 VDC

Combined Linearity& Hysteresis±1.5% FS

±1% FS±0.25% FS±0.1% FS

±0.25% FS

Operating Temperature-20°C to 80°C

-20°C to 80°C

-40°C to 150°C-40°C to 150°C-40°C to 150°C

Dimensions (mm)Range dependent

Range dependent

Range dependent

Range dependent

Range dependent

Typical Applications

Robotics, process control,
bolt clamping for bridges

Crane monitoring,offshore, loadlimited devices

Calibration presses,
robotics and effectors,
laboratory and research

Static fatigue tests,
laboratory andresearch, robotics

Assembly forces,
tool force, offshore

FN3030

FN3148FN9620

FN9630, FN9635

PackageS-beamS-beam with stops

S-beamVery high accuracy pan-cakeOperating Mode

Tension and compression

Tension and compression

Tension and compression

Tension and compression

Unique Features• Optional rod ends
• Optional high level output• Optional high compensation
temperature• Very high accuracy• High resolution• Mechanical stops• High accuracy• IP68• Entry level• High stability & accuracy• Connection flange supplied
for model FN9635• Minimal cross effect

FS Ranges

10 to 20K lbf
50 to 100K Newton

2 to 400 lbf
10 to 2,000 Newton

100 to 2,000 lbf
500 to 10K Newton

2K to 40K lbf
10K to 200K Newton

Max. Over-range

1.5X FS

5X to 100X FS

1.5X FS

3X FS

Output/Span±20 mV, 0.5-4.5 VDC±20 mV, 0.5-4.5 VDC±10 mV to ±20 mV±20 mV

Combined Linearity& Hysteresis±0.1% FS

±0.05% FS±0.05% FS±0.08% FS

Operating Temperature-40°C to 150°C-40°C to 120°C-40°C to 90°C

-40°C to 90°C

Dimensions (mm)Range dependent

Range dependent

56 x 20 x 60

Range dependent

Typical Applications

Laboratory and research, processcontrol, customized options

Product validation tests, medical
instruments, weighing

Test bed, dynamic fatigue
testing, robotics and effectors

Static fatigue tests, weighing
calibration, robotics

PAGE 12FORCE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

FORCE SENSORSFORCE LOAD CELLS
Miniature Load Cells

ELAF

XFC200R

XFL212R

XFTC300

XFU400

Package

Button, dual stud

Small diameterload button

Low profile load button

Low/high capacity
dual studMiniature rod end

Operating Mode

Tension and compression

Compression

Compression

Tension and compression

Tension and compression

Unique Features• Low cost, small profile• Microfuse technology• Low off-axis response• High stiffness• High overload capacity• Static and dynamic• Extremely flat
• Integrated load button• Small diameter• High stiffness• High overload capacity• Threaded male/
female fitting• High stiffness• High accuracy• High temperature

FS Ranges

10 to 2,000 lbf
50 to 10K Newton

0.4 to 2,000 lbf
2 to 10K Newton

1 to 100 lbf
5 to 500 Newton

0.4 to 400 lbf
2 to 2,000 Newton

100 to 1,000 lbf
500 to 5,000 Newton

Max. Over-range

2.5X FS

2X FS

2X FS

2X FS

1.5X FS

Output/Span±100 mV, 0.5-4.5 VDC±100 mV±100 mV±100 mV, 0.5-4.5 VDC±100 mV

Combined Linearity& Hysteresis±0.25% FS±0.5% FS

±1% FS±0.5% FS

±0.3% FS

Operating Temperature-40°C to 120°C-40°C to 120°C-40°C to 120°C-40°C to 150°C-20°C to 120°C

Dimensions (mm)Range dependent

Ø10 to Ø16

Ø12.5 x 3.5

Range dependent

Range dependent

Typical Applications

Theatrical rigging
loads, assembly forces,
weighing, thrustmeasurements, productvalidation testing

Material test, measuring
tools, robotics
and effectors

Dental andbiomechanical, surface
mount assembly
system, production
validation test

Material test, tool forces,
robotics end effectors

Spherical rod end
bearings, engine &suspension testing,machinery equipment

FORCE LOAD CELLSMultiaxial Load Cells

FN7110

FN7325

Package

Dual S-beam range

Custom design and ranges available upon request

Operating Mode

Tension and compression

Multiaxial force and torque

Unique Features• High resolution• Optional high level output• Double range• Measures load and torque in 3 directions, 6DOF total• Fatigue rated• Minimal cross effects

FS Ranges

2 to 2,000 lbf
10 to 10K Newton

1K to 50K lbf
5K to 250K Newton

Max. Over-range

1.2X FS

1.2X FS

Output/Span±20 mV, 0.5-4.5 VDC±150 mV, 0.5-4.5 VDC

Combined Linearity& Hysteresis±0.1% FS

±1% FS

Operating Temperature-20°C to 80°C

-20°C to 80°C

Dimensions (mm)60 x 30 x 100

Range dependent

Typical Applications

Process control, assembly forces, weighing, thrustmeasurements, product validation testing

Structure testing, crash testing, industrial test benches, robotic joints

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 13FORCE SENSORSFORCE SENSORSFORCE LOAD CELLS
Automotive Load Cells

FN2114

FN2317

FN2570

FN4055

Package

Brake pedal

Hand brake

Brake pedal

Seat belt sensor

Operating Mode

Compression

Compression

Compression

TensionUnique Features• High accuracy• Extra flat
• Compact• Easily installed• Ergonomic design• Fits most vehicles• High accuracy• Compact and extra flat
• Rugged, stainless steel design• Low operating ranges• Protected against overload• Compatible with most seat belts

FS Ranges

40 to 500 lbf
200 to 2,500 Newton

100 to 200 lbf
500 to 1,000 Newton

40 to 500 lbf
200 to 2,500 Newton

20 to 60 lbf
100 to 300 Newton

Max. Over-range

1.5X FS

1.5X FS

1.5X FS

10X FS

Output/Span±20 mV, 0.5-4.5 VDC±20 mV, 0.5-4.5 VDC±20 mV±20 mV

Combined Linearity& Hysteresis±1% FS±0.5% FS

±2.5% FS

±0.25% FS

OperatingTemperature-20°C to 80°C

-20°C to 80°C

-20°C to 80°C

-20°C to 80°C

Dimensions (mm)Range dependent

100 x 20 x 15

59 x 59 x12.5

63.5 x 63.5 x 12.7

Typical Applications

Brake pedal, clutch pedal, test bed

Hand brake, test bed

Brake pedal, clutch pedal, test bed

Auto crash testing, tension
at the belt receptacle

FN4070 & FN4080

FN7080EL20-S458

PackageSeat belt buckle sensor

Gear stick design

Special purpose seat belt load cell for automotive crash testing

Operating Mode

TensionMulti-axial

Seat belt tension

Unique Features• High operating ranges• Detachable tongue and cable• Compatible with most seat belts• Measures force inthree directions• Replaces gear knob
• Ease of mounting• Mass optimized to minimize acceleration induced
errors during SAE J2570 ATD and ISO 6487
• Optional high level and linearized outputs• Smoothed design and slotted titanium axles
eliminate drag errors and dummy damage

FS Ranges

200 to 8,000 lbf
1K to 40K Newton

40 to 100 lbf
200 to 500 Newton

1,000 to 5,000 lbf
5K to 25K Newton

Max. Over-range

1.5X FS

1.2X FS

2X FS

Output/Span±20 mV±20 mV, 0.5-4.5 VDC±20 mV, 0.5-4.5 VDC

Combined Linearity& Hysteresis±0.5% FS

< ±0.3% FS±0.5% FS

OperatingTemperature-20°C to 80°C

-20°C to 80°C

-40°C to 120°C

Dimensions (mm)Range dependent

Ø25 spherical

66 x 35 x 16.5

Typical Applications

Auto crash testing, tension
at the belt receptacle

Change gear force measurement,roughness of material

Seat belt forces, safety and restraint system crashtest, parachute tether and riser forces

PAGE 14
FORCE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

FORCE SENSORSFORCE LOAD CELLS
Digital Display Meters

ARD154

CPA150

M210

Package

Din rail mountable

Hand held indicator

Front panel or housed in caseType

Signal conditioning for wheatstonebridge sensors

Portable display suited for strain gage
type sensors

Signal conditioning and display meterNo. of Channels

Four

Two

One

Unique Features• Suited for full bridge strain gage sensors• Test stands and process industries• 2 kHz or 20 kHz max. bandwidth• Suited for 1 or 2 sensors• 45 hour life battery• Calibration pushbutton
from 0.1 to 10 mV/V

• Red LED display: ±2,000 count• High bandwidth: 1,000 Hz at -3 dB• Low noise level

Output/Span±10 VDC or 4-20 mA current output

Display only

±10 VDC

Accuracy±0.01% FS±0.005% FS±0.05% FS

Operating Temperature-10°C to 60°C

-10°C to 50°C

0°C to 50°C

Dimensions (mm)99 x 17.5 x 112

90 x 34 x 152

96 x 48 x 155

Typical Applications

Test stands, power plants, manufacturingsystems, test and measurement, test
bed regulation, automat interfaces

Outdoor punctual measurements, test andmeasurement, portable calibration device

High bandwidth test bed display,monitoring, laboratory and research,
process control equipment

M905

121

140A / 142A

Package

Front panel or housed in case

Bench top

Inline amplifier

Type

Display suited for process or
strain gage type sensors

DC amplifier and signal conditioner

DC amplifier and auto-zero

No. of Channels

One

Three

one

Unique Features• Suited for process or strain
gage type sensors• 5 digits: -19999 to 19999• Front panel programming• 0.001 to 9999
• Low noise operation with auto-zero• For bridge type sensors• µP controlled, programmable• Low pass filter options• ±1.5 mV auto-zero• For bridge type sensor (140A)• For strain gage (142A)• x10, x25, x50, x100, x200 gain
• 5 to 30 VDC excitation

Output/Span±10 VDC or 4-20 mA current output±10 VDC

0.5-4.5 VDC, ref to 2.5 VDC

Accuracy±15 bits, 20 sample/sec±0.1% FS

±0.5% FS

Operating Temperature-10°C to 60°C

0°C to 50°C

-10°C to 50°C

Dimensions (mm)96 x 48 x 60

301 x 258 x 102

56.9 x 25.4 x 12.7

Typical Applications

Display on test bed, monitoring,laboratory and researchInstrumentation labs, test benches,
R&D facilities

Instrumentation labs, test benches,
R&D facilities

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 15
FORCE SENSORSHUMIDITY SENSORSHUMIDITYSENSORS

PAGE 16
HUMIDITY SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

HUMIDITY SENSORSHUMIDITY AND TEMPERATURE (NTC) COMPONENTS
Analog Voltage and Digital Output

HS1101LF

HTU2X

HTU2XF

PackageThrough hole TO39 with side
opening plastic cap

DFN type

DFN type

Type

Capacitive humidity

Digital RH and NTC temperature

Digital RH and NTC temperature

Operating RH Range

0 to 100% RH

0 to 100% RH

0 to 100% RH

Operating Temp.

-60°C to 140°C-40°C to 125°C-40°C to 125°C

Unique Features• Robust and recognized component• Suitable for most humidity applications• Cost effective solution• Low power consumption• Fast response time
• Very low temperature coefficient• I²C interface or PWM interface or SDM interface• Low power consumption• Fast response time
• Very low temperature coefficient• I²C interface or PWM interface or SDM interface• Optimal filter

Accuracy

180 pF, ±3 pF at 55% RH±3% RH at 25°C (10 to 95% RH) ±0.3°C at 25°C

±3% RH at 25°C (10 to 95% RH) ±0.3°C at 25°C

Dimensions (mm)10 x 10 x 19

3.0 x 3.0 x 1.0

3.0 x 3.0 x 1.0

Typical Applications

Applications requiring a robust humidity
sensor in appliance, HVACR, consumer
electronics, printing, meteorology

Humidity and temperature plug and playtransducers for OEM demanding applicationsin appliance, printer, medical, HVACR

Humidity and temperature plug and playtransducers for OEM demanding applications
appliance, printer, medical, HVACRHUMIDITY AND TEMPERATURE (NTC) MINI-MODULES
Analog Voltage and Digital Output

HTU3535PVBM/Wire

HTU383X/Wire

HTG351xCH

Package

Cost effective, small size mini-module

Cost effective small size mini-module

Cost effective small size mini-module

Type

Analog voltage RH and NTC temperature

Digital RH and NTC temperature

Analog voltage RH and NTC temperature

Operating RH Range

0 to 100% RH

0 to 100% RH

0 to 100% RH

Operating Temp.

-40°C to 110°C-40°C to 110°C-40°C to 110°C

Unique Features• PTFE filter (Optional)• Electronics fully protected (5 V)• Multiple connector choices (JST, Samtec
board to board through hole)• Based on HTU21• PTFE filter (Optional)• Electronics fully protected (5 V)• Multiple connector choices (JST, Samtec
board to board through hole)• Based on HTU21• Electronics fully protected withpotting material (3.3 V or 5 V)• Multiple connector choices (JST,Samtec board to board through hole)

Calibration±3% RH at 55% RH; ±0.25°C at 25°C

±3% RH at 55% RH; ±0.25°C at 25°C

±3% RH at 55% RH; ±0.25°C at 25°C

Dimensions (mm)27 x 11.9 x YY (Depending on theconnector, from 6 to 10.8 mm length)

27 x 11.9 x YY (Depending on theconnector, from 6 to 10.8 mm length)

27 x 11.9 x 6.7

Typical Applications

Humidity and temperature plug and playtransducers for OEM demanding applications in
HVACR, home appliance, printer and medical

Humidity and temperature plug and playtransducers for OEM demanding applications in
HVACR, home appliance, printer and medical

Humidity and temperature plug andplay transducers for OEM applications in
HVACR, appliance, printer and medical

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 17
HUMIDITY SENSORSHUMIDITY SENSORSHUMIDITY AND TEMPERATURE (NTC) PROBES
Analog Output

HM1500LF

HM1520LF

HTM2500LF

Package

Probe, RH only

Probe, RH only

Probe, RH andtemperature

Type

Cost effective analog voltage RH probe

Dedicated to low RH accurate measurement

Cost effective analog voltage RH

Operating RH Range

0 to 100% RH

0 to 100% RH

0 to 100% RH

Operating Temp.

-40°C to 60°C

-40°C to 60°C

-40°C to 85°C

Unique Features• Electronics fully protectedwith potting material• Optional wiring length and connectors• Electronics fully protectedwith potting material• Optional wiring length and connectors• Electronics fully protectedwith potting material• Optional wiring length and connectors

Calibration±3% RH at 55% RH±3% RH at 10% RH±3% RH at 55% RH; ±0.25°C at 25°C

Dimensions (mm)57 x 11 x 11 (Standard wire length of 200 mm)

57 x 11.5 x 11.5 (Standard wire
length of 200 mm)

86 x 11.5 x 11.5 (Standard wire
length of 200 mm)

Typical Applications

Medical, telecommunication cabinets,
green houses, process control, industrial

Medical, drying cabinets, lowhumidity, meteorology

Hygrostat, data loggers, cabinets

PAGE 18
HUMIDITY SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

HUMIDITY SENSORSHUMIDITY AND TEMPERATURE (NTC) SENSORSFrequency Output Systems (Digital)

HTF3000LF
Package

PCB for board to board

Type

Frequency output for RH, direct NTC for temperature

Operating RH Range

0 to 100% RH

Operating Temp.

-40°C to 85°C

Unique Features• Voltage supply from 3 to 8 VDC• Through hole or SMD• T and R available

Calibration±3% RH at 55% RH; ±0.25°C at 25°C

Dimensions (mm)12.5 x 18.5 x 11.2

Typical Applications

HVACR, printer, cabin comfort, hygrostat

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 19
HUMIDITY SENSORSLIQUID LEVEL SENSORSLIQUID LEVELSENSORS

PAGE 20LIQUID LEVEL SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

LIQUID LEVEL SENSORSLIQUID LEVEL SWITCHES
Side Entry

LS304-31

LS509-51

LS809-31

LDS309-11N

LCS-03

RCS01-10

Package

Glass filled nylon 6.6

Glass filled PPS

Glass filled
polypropylene

Glass filled nylon 6.6

Acetal/Polypropylene

Polypropylene

Type

Level sensor

Level sensor

Level sensor

Level sensor

Level sensor

Level sensor

Unique Features

1/2" NPT horizontal
mount SPDT

M16 horizontal
mount SPST

1/2" NPT horizontal
mount SPST

M16 horizontal
mount SPST

Push fit horizontal
mount SPST

M16 Horiz
mount SPST

Max. Pressure

4.7 bar

4.7 bar

2.0 bar

4.7 bar0.34 bar

4.7 barOperating Temp.

-30°C to 130°C-30°C to 110°C-30°C to 105°C-30°C to 130°C-30°C to 60°C

-30°C to 70°C

Dimensions (mm)103 x 29 x 29

88 x 27 x 27

103 x 29 x 29

100 x 27 x 27

100 x 36 x 36

67 x 30 x 30

Typical Applications

Chemical high or low
level, diesel fuel, fuellow level, alcohols,
low oil detection

Coolant levelindication, waterhigh or low level,boiler heating
element protection,
drinking water level,boiling waterContinuous 80°C
in water, waterhigh or low level,
condensate level
alarm, drinking waterlevel, cooling systems

Chemical high or low
level, diesel fuel, fuellow level, alcohols,
low oil detection

Water high or low,
unpressurized central
heating systems, fuellevel, organic solvents

Waste water level,coolant, water

LIQUID LEVEL SWITCHES
Top & Bottom Entry

VS309-51N

VS804-21

VCS-04

VCS-06

EVS312-11N

EVS722-51

Package

Glass filled nylon 6.6

Glass filled
polypropylene

Polypropylene

Nylon 6.6

Nylon 6.6

Brass/ Polypropylene

Type

Level sensor

Level sensor

Level sensor

Level sensor

Level sensor

Level sensor

Unique Features

M16 verticalmount SPST

M16 verticalmount SPDT

M8 verticalmount SPST

M8 verticalmount SPST

M16 vertical mount
2 X SPST

M16 vertical mount
16 X SPST

Max. Pressure

4.7 bar

4.7 bar

1.3 bar

1.3 bar

4.7 bar

4.7 barOperating Temp.

-30°C to 130°C-30°C to 105°C-30°C to 70°C

-30°C to 105°C-30°C to 110°C-30°C to 80°C

Dimensions (mm)87 x 22 x 22

87 x 22 x 22

59 x 22 x 22

59 x 22 x 22

224 x 22 x 22

224 x 22 x 22

Typical Applications

Chemical high or low
level, diesel fuel, fuellow level, alcohols,
low oil detection

Continuous 80°C
in water, waterhigh or low level,
condensate level
alarm, drinking waterlevel, cooling systems

Water high or low,
condensate level,
cooling sytems

Chemical high or low
level, diesel fuel, fuellow level, alcohols,
low oil detection

Chemical high or low
level, diesel fuel, fuellow level, alcohols,
low oil detection

Coolant levelindication, water highor low level, boiler
heating element
protection, sump level

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 21LIQUID LEVEL SENSORSPHOTO OPTIC SENSORSPHOTO OPTICSENSORS

PAGE 22
PHOTO OPTIC SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

PHOTO OPTIC SENSORSPHOTO OPTIC SENSORS
Photo Optic Components

ELM-4000

EPM-4001

ELM-5000

EPM-5000

Package

Lead frame

Lead frame

Surface mountSurface mountType

Emitter assembly

Detector assembly

Emitter assembly

Detector assembly

Range

660 nm / 880-940 nm—

660 nm / 890-905 nm—

Unique Features• Low cost
• Dual drive• Clear epoxy lens

• Low cost
• Fast response• High efficiency• Reflow solderable• Dual drive• Clear epoxy lens• Reflow solderable• Fast response• High efficiency

Accuracy

Sensor dependent

Sensor dependent

Sensor dependent

Sensor dependent

Operating Temp.

-20°C to 80°C

-20°C to 80°C

-20°C to 80°C

-20°C to 80°C

Dimensions (mm)4.4 x 5.1 x 1.9

4.4 x 5.1 x 1.8

4.0 x 4.8 x 1.3

4.0 x 4.8 x 1.3

Typical Applications

Pulse oximetry, finger andear probes, disposable

Pulse oximetry, finger andear probes, disposable

Pulse oximetry, finger andear probes, disposable

Pulse oximetry, finger andear probes, disposable

PHOTO OPTIC SENSORS

Pulse Oximetry (SpO2) Probe Platforms

Disposable Sensor

Finger Clip Sensor

Soft Sensor

Package

Biocompatible

Biocompatible

Silicon bootType

Sensor platform

Sensor platform

Sensor platform

Range

Adult/neonatal

Adult

Adult/pediatricUnique Features• Latex free
• Lightweight• Microfoam/cloth• Soft pads
• Lightweight• Easily cleaned• Ease of use• Lightweight• Latex free

Accuracy

Sensor dependent

Sensor dependent

Sensor dependent

Operating Temp.

-20°C to 80°C

-20°C to 80°C

-20°C to 80°C

Typical Applications

Pulse oximetry

Pulse oximetry

Pulse oximetry

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 23
PHOTO OPTIC SENSORSPIEZO FILM SENSORSPIEZO FILMSENSORSPIEZO FILM

DT1, SDT1

Piezo Cable

CM-01

FLDT1

Package

Unshielded element with
twisted pair or shielded
element with shielded cable

Shielded coaxial 20
gage piezo cable

Metallized plastic housing

Unshielded film elementwith screen printed leads

Type

Flexible film, adhesive mountPolymer jacketing,armored jacketing

Contact microphone

Flexible film, adhesive mountRange

15 mV/µε up to 1% strainµPa sensitivity

40 V/mm; 8 Hz to 2.2 kHz

15 mV/µε, up to 1% strain

Unique Features• Thin, flexible, robust• Withstands >2% strain• Ultra-low power (Selfgenerating)• Continuous lengthsof up to 1 km
• Shielded construction• Low noise• Shielded construction• High sensitivity• Thin, flexible• Leads screen printed on film
• Connects to standard connector

Accuracy±20% (Typical)±20% (Typical)—±20% (Typical)

Operating Temp.

-40°C to 70°C
(Higher available custom)-40°C to 85°C

5°C to 60°C

-40°C to 70°C;(Higher available custom)

Dimensions (mm)Application dependentØ3 (Continuous lengths)

Ø18 x 11 high

12 x 30 active; (Custom available)

Typical Applications

Dynamic strain gage, contact
microphone, acoustic pickup

Perimeter and fence security,geophone, impact sensors,
intrusion detection, seatoccupancy (e.g. airbag), patient
bed vital signs monitor

Electronic stethoscope, contact
microphone, vibration

Event timing, dynamic strain,
motion detection

PAGE 24PIEZO FILM SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

PIEZO FILM SENSORSPIEZO FILM

Sleep Monitor Strip

BL Traffic Sensor

Laboratory Amplifier

80 KHz Transducers

Package

Unshielded element with crimps

Center Core: 16 gage copper wire
Piezoelectric Material:
Piezoelectric film cableOuter Sheath: 0.016” thick brass

Bench top

Pin mounted

Type

Flexible film, adhesive mountSpiral wrapped PVDF
piezo film cable

Piezo film lab amp

Air ultrasound transducer

Range

15 mV/µε up to 1% strain

15 mV/µε up to 1% strain

0.1 Hz to 100 kHz

80 kHzUnique Features• Thin, flexible, robust• Withstands >2% strain• Ultra-low power (Selfgenerating)• Flexible, durable, available
in many lengths• Withstands >2% strain• Ultra-low power (Selfgenerating)• Voltage or charge
mode settings• Multi-pole high-passand low-pass filters• Adjustable gain

• Small size
• Low mechanical Q• Shielded package

Accuracy±20% (Typical)±20% (Typical)

Application dependent

Application dependent

Operating Temp.

-40°C to 70°C
(Higher available custom)-40°C to 70°C
(Higher available custom)

0°C to 40°C

-20°C to 80°C

Dimensions (mm)28 um PVDF; 8mm x 800mm

0.260” wide x 0.063” thick;0.005"

150 x 100 x 100

Ø6 x 9

Typical Applications

Respiration and heart beatmonitoring for mattress or seat

Traffic counting, classifying,toll booths, speed detection,
red light cameras

Low frequency dynamic strain,
pyroelectric signals, machine
vibration, piezo cable andtraffic sensor interface

Air ranging, ultrasonicmouse, digitizers

NDT-1

Tamper Box

ACH-01LDTC Family

Package

Adhesive mounted

Flat film or box mounted

Ceramic base, plastic
cover, shielded cable

Piezo film elements with orwithout mass

Type

High frequency ultrasoundtransducer

Tamper detection sensor

Adhesive mountCantilever beam withvertical or horizontal pins

Range

3 MHz

Application dependent±250 g (Typical)±10 g (Typical)

Unique Features• Flexible• High bandwidth, low Q• Low impedance• Low power• Custom shapes and sizes• High security• Extremely high bandwidth• Low cost
• Ultra-low power• Very low cost
• High sensitivity (1 V/g)
• Ultra-low power(Self generating)Accuracy

Application dependent

Application dependent±20% (Typical)±20% (Typical)

Operating Temp.

-20°C to 60°C

-40°C to 85°C

-40°C to 85°C

-40°C to 70°C

Dimensions (mm)12 x 30

Application dependent

18.80 x 13.21 x 6.10

19.05 x 6.35 x 6.35

Typical Applications

Thickness measurement,
speed of sound measurement,
pulse/echo NDT

Encryption modules, POS card
readers, PIN entry devices

Vibration sensing, gear box andhigh speed monitoring, highspeed bearings and centrifuges,
speaker motional feedback

Wake-up switch, load imbalance,
antitheft devices, impact
sensing, vital signs monitoring

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 25
PIEZO FILM SENSORSPOSITION SENSORSPOSITIONSENSORS

PAGE 26POSITION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

POSITION SENSORSANISOTROPIC MAGNETORESISTIVE (AMR) SENSOR COMPONENTS
Magnetoresistive (MR)KMY, KMZMS32

KMT39 (Former 32B), KMT37

Package

SOT-223, E-line 4 pin

TDFN

TDFN

Type

Linear low field sensor

Low field switch sensor

Angle sensor

Range-2 to 2 kA/m magnetic field

1 to 3 kA/m magnetic switching field

180° angle

Unique Features• High sensitivity• Low hysteresis• Linear to uniaxial field strength• Linearized ratiometric output• Temperature compensated switching point• High accuracy• High resolution

OutputRatiometric with output
voltage range 20 mV/V

Ratiometric with output
voltage range 10 mV/V

Sine and cosine signals with output
voltage range 20 mV/V

Resolution

Typ. 0.1% of range

Typ. 0.1 kA/m

Typ. 0.01° to 0.1°

Accuracy

Typ. 1.0% of range

Typ. 0.1 kA/m

Typ. 0.1° to 1.0°

Operating Temp.

-40°C to 150°C-25°C to 85°C

-40°C to 150°C(175°C on request)

Dimensions (mm)SOT: 6.6 x 7.0 x 1.6 E-line: 16 x 4.2 x 2.4

TDFN: 2.5 x 2.5 x 0.8

TDFN: 2.5 x 2.5 x 0.8 SO-8: 5 x 4 x 1.75

Typical Applications

Non-destructive material testing, sprayarm detection in dish washers, magnetic
imaging, brake pedal position

Piston position switch, reedswitch replacement

Steering position, flow meters,
rpm meters, rotary encoders

KMT36H

KMXP

KMA36

Package

TDFN 2.5 x 2.5

DFN 2 x 6

TSSOP

Type

Angle sensor

Linear displacement sensor, 3 flat and3 perpendicular versions

Angle sensor with digital output

Range

360° angle

Absolute within magnetic pole pitch,else incremental

360° angle

Unique Features• High accuracy• High resolution• 360° full turn

• For pole pitch• KMXP 1000: p= 1 mm
• KMXP 2000: p= 2 mm
• KMXP 5000: p= 5 mm

• Low cost MR encoder for rotational
and incremental measurements• Digital output

OutputThree 120° phase shifted output signals
with output voltage range 20 mV/V

Sine and cosine signals with output
voltage range 20 mV/V

Analog and I2C Digital

Resolution

Typ. 0.01° to 0.1°

0.01% to 0.1% of pole pitch

Typ. 0.1°

Accuracy

Typ. 0.1° to 1°

0.1% to 1.0% of pole pitch

Typ. 0.3°

Operating Temp.

-40°C to 150°C-40°C to 125°C-25°C to 85°C

Dimensions (mm)

TDFN: 2.5 x 2.5 x 0.8

DFN: 2 x 6 x 0.8

TSSOP20: 6.5 x 6.4 x 1.2

Typical Applications

Steering position, gage readings, rotary
encoders

Roller conveyors, circular saws, bending
machines

Knobs, small robotics, angular/linear position

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 27
POSITION SENSORSPOSITION SENSORSANGULAR POSITION TRANSDUCERS—INDUCTIVE
Absolute

RVIT-Z

R60D

R30A

Package

PCB for OEM volumes

Servo mount with ball bearing

Servo mount with ball bearing

Resolution

Infinite

Infinite

Infinite

Excitation

DC voltage

DC symmetrical ±15 VDC

AC operated

OutputDC voltage, DC current, digital±7.5 VDC

AC voltage

Range

Up to ±75°

±60°±30° to ±60°

Unique Features• Absolute position• Absolute position• Low momentum of inertia• Absolute position

Operating Temp.

-25°C to 85°C

-25°C to 85°C

-55°C to 150°C

Dimensions (mm)CustomAluminum case size 11 (Ø27 mm)

Aluminum case size 11 (Ø27 mm)

Typical Applications

Viscometers, valve position, robotics,
HVACR vane position, ATM’s, joysticks

Dancer arm position, rotary actuator
position feedback, throttle lever position
feedback, ball valve position, textile
manufacturing equipment, printing presses

Machine tool equipment, rotaryactuator feedback, valve positioning,
power generation valve position

ANGULAR POSITION—ENCODERS
Absolute

H005, H009

H009, 1200

R36

Package• 12.7 mm - 22.19 mm/500 in .875 in housing diameter• 3.170 mm/.1248 in shaft diameter• 16.9 mm - 17.4 mm/.670 in .680 in housing length• 22.23 mm/.875 in housing diameter• 3.170 mm/.1248 in shaft diameter• 26.1 mm/1.03 in housing length

Heavy duty shaftlessRange

Up to 359°

Up to 359° (Dual output)

180°

Output Options

Analog/PWM/Serial

Analog/PWM/Serial

Voltage

Resolution

12-bit analog/PWM 14-bit serial (SPI)

12-bit analog/PWM 14-bit serial (SPI)

Analog 1.4°

Excitation

—

—

5 VDC

Unique Features——• Rugged housing• Shaftless• No optical degradation

Absolute Linearity±0.2%± 0.2% (Dual output)—

Nominal Supply

5 volts

5 volts (Dual output)—

Operating Temp.

-40°C to 150°C-40°C to 150°C-40°C to 85°C

Rotational Life

> 100 million cycles (Bearing life)> 100 million cycles (Bearing life)—

Critical position feedback applications
in commercial, industrial, medical,
aircraft and military markets

Critical position feedback applications
in commercial, industrial, medical,
aircraft and military markets

Dual Output

Dimensions (mm)Typical Applications

PAGE 28
POSITION SENSORS

37.36 x 25.4 x 7.62

te.com/sensors

Feedback sensor or human machineinterface device, rudder control,
servomotor position and speed control

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

POSITION SENSORSTILT SENSORSSingle Axis

E-Series

AccuStar EA

APS System

Package

Ceramic housing

LCP housing

Plastic housing

Type

Inclination sensor module

Inclinometer sensor module

Inclination systemRange±5°, ±15°±45° to ±60°±45°, ±90°

OutputVoltage

Voltage

Analog/digital

Unique Features• Easy to handle• Minimal temperature drift• Good long term stability• Compact• Low power• Vertical and horizontal mount• Stand alone system• Separate system and sensor

Accuracy±0.2° to ±0.5°

0° to 10° ±0.1% accuracy
10° to 60° ±0.75% reading

0° to 10° ±0.1% accuracy
10° to 45° ±0.75% of reading

Operating Temp.

-25°C to 85°C

-30°C to 65°C

-25°C to 65°C

Dimensions (mm)29 x 17 x 16.5

65.91 x 51.56 x 30.5

127.5 x 88 x 32.2

Typical Applications

Road construction, building monitoring,weighing systems, mobile andstationary cranes, platform leveling

Wheel alignment, construction, equipment,
antenna positioning, robotics,
crane/boom angle

Tower crane safety, RV and mobile trailer
leveling, water and oil well drilling rigs,
mining equipment

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 29POSITION SENSORSPOSITION SENSORSTILT SENSORSSingle Axis

G-Series

IT9000

AccuStar IP66

Package

Aluminum housing IP67

Aluminum or stainlessAluminum housing IP66

Type

Inclinometer

Inclinometer

Inclinometer

Range

±10°

±45° to ±240°±3° to ±45°

OutputSwitchVoltage divider, 4 - 20 mA

Current

Unique Features• Programmable• EMC standard• High switch accuracy• Rugged industrial design, IP67/68• Submersible• Designed for brutal environments• CSA, CENELEC certification for
hazardous area applications• EMI and RFI rated• CE pending• Water tight enclosure

Accuracy±0.25°±1%

0° to 10° ±0.1% linearity
10° to 45° ±1% linearity

Operating Temp.

-25°C to 85°C

-34°C to 90°C

-25°C to 60°C

Dimensions (mm)80 x 75 x 57.5

Ø130 x 100

98.04 x 63 x 35.05

Typical Applications

Lift platforms, building device control, train
inclination monitoring, position switch

Waste water control, tainter gates, draw
bridges, heavy industrial applications

Tower crane safety, RV and mobile trailer
leveling, water and oil well drilling rigs,
mining equipment

TILT SENSORS
Dual Axis

DPL, DPN

DOG2

DPG

D-Series

Package

PCB board

Plastic PA 6.6 housing, IP67

Aluminum housing IP67

Aluminum housing IP67

Type

Inclination board module

Inclinometer

Inclinometer

Inclinometer

Range±2° to ±30°±25°, ±45°, ±90°±5° to ±30°

±5° to ±30°OutputVoltage/RS 232/SPI

Voltage/Current/
J1939/CANopen®

RS232/Voltage

RS232/Voltage/Current/
Switch/PWM/CANopen®

Unique Features• High resolution• Minimal temperature drift• User configurable• Plug and play
• Wide measurement range• Cost-efficient• Cable with connector• Fast MEMS sensor• CE approved• Rugged housing• Easy to use• User configurable• High accuracy• Rugged housing• Programmable• CE approved

Accuracy±0.05° to ±0.8°± 0.5° (Full temp. range)±0.05° to ±0.3°±0.04° to ±0.8°

Operating Temp.

-40°C to 85°C

-40°C to 85°C

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)45 x 45 x 20

70.5 x 45 x 15

84 x 70 x 34.2

84 x 70 x 46

Typical Applications

Laser leveling, weighing systems,
mobile and stationary cranes,
hydraulic leveling, building
monitoring, wind power

Off road vehicle, fork lift, truck
leveling, man lift, harvester, farmmachine, tip over protection,
solar panel control

Platform leveling, road
construction machines, tunnel
drilling, mobile leveling

Drilling machines, mobile andstationary cranes, wind power,
antenna/radar leveling

PAGE 30POSITION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

POSITION SENSORSPROXIMITY SENSORS

PS801

PS811

PS831

PS2011AB

Package

Stainless steel

Nylon 6.6

Stainless steel

Glass filled nylon 6.6

Type• Proximity sensor• Used with proximity magnet• Proximity sensor• Used with proximity magnet• Proximity sensor• Used with proximity magnet• Proximity sensor• Used with proximity magnet

Unique Features

SPST reed switch, normally open

SPST reed switch, normally open

SPST reed switch, normally open

SPST reed switch, normally open

Operating Temp.

-30°C to 120°C-30°C to 110°C-30°C to 130°C-30°C to 105°C

Dimensions (mm)Ø12 x 65

Ø10 x 38

Ø12 x 32

29 x 7 x 20

Typical Applications

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

PS2021AB

PS2031AB

PS501

Package

Glass filled nylon 6.6

Glass filled nylon 6.6

Glass filled nylon 6.6

Type• Proximity sensor• Used with proximity magnet• Proximity sensor• Used with proximity magnet• Proximity sensor• Used with proximity magnet

Unique Features

SPST reed switch, normally closed

SPDT reed switch

SPST reed switch, normally open

Operating Temp.

-30°C to 105°C-30°C to 105°C-30°C to 130°C

Dimensions (mm)29 x 7 x 20

29 x 7 x 20

Ø6 x 32

Typical Applications

Door interlocks, hook switches, security
systems, safety interlocks, position indication

Door interlocks, hook switches, security
systems, safety interlocks, position indication

Door interlocks, hook switches, security
systems, safety interlocks, position indication

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 31POSITION SENSORSPOSITION SENSORSPROXIMITY MAGNET

PM101

PM50

PM81

PM83

Package

Glass filled nylon 6.6

Glass filled nylon 6.6

Nylon 6.6

Stainless steel

Type• Proximity magnet• Used with proximity sensor• Proximity magnet• Used with proximity sensor• Proximity magnet• Used with proximity sensor• Proximity magnet• Used with proximity sensor

Unique Features

Housed magnet

Housed magnet

Housed magnet

Housed magnet

Operating Temp.

-30°C to 105°C-30°C to 70°C

-30°C to 120°C-30°C to 120°C

Dimensions (mm)29 x 7 x 20

Ø6 x 32

Ø10 x 38

Ø12 x 32

Typical Applications

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

Door interlocks, hook switches,
security systems, safety
interlocks, position indication

LINEAR POSITION TRANSDUCERS
Cable Extension Transducers

PT1, PT5

PT8000

PT9000Range

0 - 2 to 0 - 250 inches

0 - 2 to 0 - 60 inches

0 - 75 to 0 - 1700 inches

OutputVoltage divider, 0 - 5 VDC, 0 - 10 VDC,
4 - 20 mA, incremental encoder,
CANbus, DeviceNet™, RS-232

Voltage divider, 0 - 5 VDC, 0 - 10 VDC,
4 - 20 mA, incremental/absolute encoder,
CANbus, DeviceNet™, RS-232

Voltage divider, 0 - 5 VDC, 0 - 10 VDC,
4 - 20 mA, incremental/absolute encoder,
CANbus, DeviceNet™, RS-232

IP Rating

IP65, IP67 (PT5)

IP67, IP68

IP67, IP68

Enclosure

Aluminum and abs plastic (PT1)

Aluminum or stainlessAluminum or stainless

Accuracy±0.04% to ±0.25%±0.04% to ±0.25%±0.04% to ±0.25%

Unique Features• Designed for most factory environments• Industry standard output signals• User serviceable• Compact design (PT1)• Heavy duty, submersible• Designed for extreme industrial
and marine environments• CSA, CENELEC certification for
hazardous area applications• High accuracy, high acceleration• Free-release proof with VLS option• M12 and DEUTSCH connector options• Heavy duty, submersible• Proven workhorse for longstroke applications• Designed for extreme industrial
and marine environments• CSA, CENELEC certification for
hazardous area applications• Free-release proof with VLS option• M12 and DEUTSCH connector options

Operating Temp.

-40°C to 90°C

-40°C to 90°C

-40°C to 90°C

Dimensions (mm)85 x 100 x 70 (PT1) 100 x 175 x 80 (PT5)

90 x 140 x 135

200 x 135 x 125

Typical Applications

Factory automation, industrial,
die casting, injection molding

Steel mills, lumber and paper mills, factory
automation, die-casting, injection molding,mobile construction and mining

Mobile hydraulic boom position, water
resource management, mining andtunnel boring equipment, telescoping
mechanism position, theater stage control

PAGE 32POSITION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

POSITION SENSORSLINEAR POSITION TRANSDUCERS
Cable Extension Transducers

M150, MTA

MT2, MT3

SM, SPRange

0 - 1.5 to 0 - 5 inches

0 - 3 to 0 - 30 inches

0 - 2.5 to 0 - 50 inches

OutputVoltage divider

Voltage divider, incremental encoder

Voltage divider, 0 - 10 VDC, 4 - 20 mA

Environment/
IP Rating

IP50

IP50, IP67 (MT3A)

IP50, IP67 (SP)

Enclosure

Aluminum

Aluminum and polycarbonate

Polycarbonate with stainless steel bracket

Accuracy±0.4% to ±1%±0.25% to ±1.1%±0.25% to ±1%

Unique Features• M150: one of the world’s smallest
string potentiometer• Designed for space-critical and testing
applications• Designed for test applications• Dual-axis measuring cable alignment• Tracks high-acceleration linear
position up to 136g’s• High-frequency response• GAM EG 13 certification• Compact design• M12 connection• Adjustable mounting bracket• Free-release tolerant• Custom configurations for OEMS

Operating Temp.

-40°C to 85°C (M150)-55°C to 100°C (MTA)-55°C to 125°C-18°C to 70°C (SM)-40°C to 85°C (SP)Dimensions (mm)19 x 19 x 10 (M150)

55 x 45 x 55

120 x 140 x 140

Typical Applications

Aerospace, automotive instrumentation,
automotive crash testing, automotive
and motorcycle racing

Automotive crash testing, aerospace andflight testing

Factory automation, light industrial, seismic
testing, racing instrumentation, medical
imaging systems, fume hood position

SG, SRSK

PTX, PT101

Range

0 - 80 to 0 - 175 inches

0 - 250 and 0 - 400 inches

0 - 2 to 0 - 100 inches

OutputVoltage divider, 0 - 5 VDC, 0 - 10 VDC,
4 - 20 mA, incremental encoder, CANbus

4 - 20 mA, 0 - 10 V, voltage divider,
CAN J1939, CANopen®, Encoder drive

Voltage divider, 0 - 5 VDC, 0 - 10 VDC,
4 - 20 mA, incremental encoder,velocity output (DV301)

Environment/
IP Rating

IP67

IP67

IP50

Enclosure

Polycarbonate with stainless steel bracket

Polycarbonate with stainless steel bracket

Aluminum

Accuracy±0.35% to ±0.5%±.25% FS

±0.04% to ±0.25%

Unique Features• Low cost, high value string potentiometer• Versatile stainless steel mounting bracket• Simple one-button user scalable
stroke range (SR)• Custom configurations available
for OEM customers• Compact design• M12 connectivity• Adjustable mounting bracket• Original classic design• High precision• Proven track record

Operating Temp.

-40°C to 85°C

-40°C to 85°C

-40°C to 90°C

Dimensions (mm)100 x 120 x 200

120 x 140 x 140

Model and range specific

Typical Applications

Outdoor mobile construction equipment,
outrigger positioning, hydraulic lifts, waterand power controls

Mobile construction equipment, factory
automation

Aerospace testing, architectural andstructural testing, factory automation

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 33POSITION SENSORSPOSITION SENSORSLINEAR POSITION TRANSDUCERS—INDUCTIVE
Absolute

HR

M12

HC

Package

AISI-400 series stainless steel

AISI-304 series stainless steel

AISI-400 series stainless steel

Linearity±0.25% of range±0.25% of range±0.25% of range

Excitation

AC operated

AC operated

AC and DC operated versions

OutputAC voltage

AC voltage

AC or DC voltage, 4 - 20 mA loop or RS-485

Range±0.05 to ±10 inches±10 to ±100 mm±0.05 to ±10 inches

Unique Features• Large bore to core clearance• Broad range of excitation frequencies• Variety of options• Mild radiation resistance option• Metric series• High stroke to length ratio• Constant sum of secondaries• Excellent temperature coefficient• Hermetically sealed• Welded connector• Double shielding• Intrinsically safe version• CE mark for DC versions

Operating Temp.

-55°C to 150°C (220°C optional)-55°C to 150°C (220°C optional)-55°C to 150°C (AC); 0°C to 70°C (DC)

Diameter (mm)

20.6

12

19

Typical Applications

General industrial

Hydraulic spool valve position feedback, flight
simulators, aircraft flight control feedback

Harsh environments, submersible
applications, process controls,
valve position feedback

XS-C

DC-SE

XS-D

Package

AISI-304 series stainless steel

AISI-400 series stainless steel

AISI-400 series stainless steel

Linearity±0.25% of range±0.25% of range±2% of range

Excitation

AC operated

8.5 to 28 VDC

AC operated

OutputAC voltage

0 - 5 VDC (4 wire), 1 - 6 VDC (3 wire)

AC voltage

Range±0.25, ±0.5 and ±1 inches

0 - 0.1 to 0 - 6 inches±1 to ±10 inches

Unique Features• High pressure• Bulkhead mounting• Hermetically sealed welded assembly• CE mark• Low current consumption (6 mA typical)• Synchronous demodulation• Shielded cable• Very high stroke to body length ratio

Operating Temp.

-55°C to 150°C-25°C to 85°C

-55°C to 150°C

Diameter (mm)

19

19

20.6

Typical Applications

Hydraulic actuators, other pressurized vessels

Positioning sensing feedback,battery operated systems, test labs,
ram guide, platen position

Where sensor installation length is restricted,
ideal replacement for linear potentiometers

Other models available, please consult TE.com.

PAGE 34POSITION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

POSITION SENSORSLINEAR POSITION TRANSDUCERS—INDUCTIVE
Absolute

MACRO HSTA/R

MACRO SSI/R

MACRO HPGS 750

Package

AISI-410 stainless steel

Alloy 625

AISI-410 stainless steel

Linearity±0.25% of range±0.10% of range±0.25% of range

Excitation

AC operated

AC or DC operated

AC operated

OutputAC voltage

AC or 4-20 mA loop digital CANbus available

AC voltage

Range±0.050 to ±10.0 inches±1.0 to ±10.0 inches±0.050 to ±10.0 inches

Unique Features• IP68 rating, hermetically sealed• Mild radiation resistant (30 Mrad) optional• Axial or radial connector with thru-bore
construction• Operating pressure to 5,000 psi• (7,500 psi proof)• Seawater submersible IP68
• Standard Seacon connector• Axial or radial connection• Radial screw-on 38999 connector• IP68 rating, hermetically sealed• Designed for high vibration applications

Operating Temp.

-55°C to 200°C standard (Contact
factory for higher temperature)-40°C to 80°C

-55°C to 200°C

Diameter (mm)

19

23.9

19

Typical Applications

High temperature steam and gas
valves, nuclear power plants, harsh andcorrosive environments, environmentswith heavy dust, dirt, and humidity

Off-shore drilling platforms, pipeline
monitoring, choke valves, mooring cables,
extensometers, pulp and paper mills

Nuclear power generation equipment, hydrauliccylinder position, steam valve positioning,
power generation equipment, corrosive
environments, high-vibration environments

MACRO CD375

MACRO GHSE/R

Package

AISI-410 stainless steel

AISI-410 stainless steel

Linearity±0.25% of range±0.1% of range

Excitation

AC operated

DC operated

OutputAC voltage

0 - 10 VDC

Range± 0.025 to ±1 inches

0.100 to 4 inches

Unique Features• Compact design• Operating pressure to 20,000 psi+

• Spring loaded design• IP68 rating, hermetically sealed• Axial and radial connection• Low pressure air-extend/spring-retract
version available (GHSER 750-A)

Operating Temp.

-55°C to 200°C-20°C to 70°C

Diameter (mm)

9.5

19

Typical Applications

Machine tools, robotic grippers, medical equipment, valve position
sensing, hydraulic cylinder, down-hole equipment

Industrial gaging systems, replaces dial indicators,
fabricated metal products gaging

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 35POSITION SENSORSPOSITION SENSORSLINEAR POSITION TRANSDUCERS—INDUCTIVE
Dimensional Gaging Products

LBB Spring-Extend

LBB Air-Extend

PCA 375

GC

Ultimate-Precision
Digital LBB

Linearity±0.2% of range±0.2% of range±0.5% of range±0.25% (Voltage) to ±0.5%(4 - 20 mA) of range

Accuracy ±0.2%

Excitation

AC operated

AC operated

AC operated

AC or DC voltage

5 VDC USB (Bus orexternal)

OutputAC voltage

AC voltage

AC voltage

AC or DC voltage, RS485, or 4 - 20 mA loop

RS485; USB

Range±0.02 to ±0.20 inches±0.04 and ±0.1 inches±0.02 to ±1 inches±0.05 to ±2 inches

1, 2, 5 and 10 mm

Unique Features• 0.000004 inch (0.1µm) repeatability• Removable tungsten
carbide contact tip• Double shielded LVDT• Repairable• 0.000004 inch (0.1µm) repeatability• Removable tungsten
carbide contact tip• Double shielded LVDT• Repairable• Longer strokes• IP65 cable exit
• Accepts industry
standard contact tips
• Heavy duty return spring• Hermetically sealed• Welded MS connector(MIL-C-5015) "• CE mark for DC versions• Special tips available• Air extend spring
retract available• Plug-and-play• 14-bit resolution• COM libraries provided• CE mark• USB adapter and power
supply available

Operating Temp.

-40°C to 70°C

-40°C to 70°C

-20°C to 70°C

-55°C to 150°C (AC)0°C to 70°C (DC)0°C to 60°C

Diameter (mm)8 or 9.5

8 or 9.5

9.5

19 mm body, 1/2- 20 threads

Stackable gage system

Typical Applications

Process standards,
manufacturing on-line
inspection, robotics,
replaces dial indicators
in manual measurement
systems

Process standards,
manufacturing on-line
inspection, robotics,
replaces dial indicators
in manual measurement
systems

High density gaging
fixtures, resistance weld
verification, pressing
applications, X-Y stage
position feedback, roughcasting inspection

Harsh environments,
environments requiring
hermetic seal, hightemperatures (150°C
for AC units)

Multi-channel electronic
dimensional gaging,
precision dimensional
measurement, optics
inspection systems, SPC
data collection, hand tools

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

LINEAR POSITION ENCODERS
Incremental

ED32i
Package

IP67 aluminum

Range

Magnetic scale, 5 mm pole pitch, typically up to 100 m
absolute version up to 100 mm range on request

Excitation

5 VDC

Output5 V TTL ABZ differential quadrature; RS-485

Resolution≥10 µm; field programmable

Max. Speed

4 m/s

Unique Features• Contactless incremental measurement• Very high accuracy, programmable resolution• High speed up to 4 m/s• Error detection, missing scale function• Adapter plate for easy mounting

Operating Temp.

-25°C to 85°C

Dimensions (mm)60 x 20 x 10

Typical Applications

Linear displacement measurement inindustrial and medical applications

PAGE 36POSITION SENSORS

te.com/sensors

POSITION SENSORSLVDT/RVDT INSTRUMENTATION

LVM-110, LiM-420

LDM-1000

ATA-2001

Package

Open circuit board

DIN rail mount

1/8 DIN panel mount

SupplyDC voltage

10 to 30 VDC

115 and 220 VAC,
50 - 400 Hz

OutputDC voltage or current

DC voltage and current

DC voltage and current

Operating Temp.

0°C to 55°C

-25°C to 85°C

-40°C to 85°C

Unique Features• Master/slave for multi-up applications• Dip switch selectable excitation frequencies• Plug-in PCB or wire termination• Small form factor• Operates with 4, 5 & 6 wire LVDT/RVDTs• Adjustable zero, span and phase• Status LEDs
• CE mark• Push button programmable• Splash proof front panel• LED status lights• Mounting hardware included• CE markDimensions (mm)63 x 56 x 21

115 x 99 x 23

267 x 99 x 49

Typical Applications

OEM applications

Automotive test track instrumentation,
gas and steam turbine controls,
factory automation

Precision metrology labs, power
generation valve position monitoring

PML 1000

MP 2000

MMX Mini Module

Package

1/8 DIN panel mount

1/4 DIN panel mount

DIN rail mount

Supply90 to 265 VAC,
50 - 60 Hz or 24 VDC

100 to 240 VAC,
47 - 63 Hz

15 to 30 VDC

OutputDC voltage and current (RS-485 optional)

DC voltage andRS-232

DC voltage or 4-20 mA

Operating Temp.

10°C to 55°C

0°C to 55°C

0°C to 70 °C

Unique Features• 5 digit LED display• Auto-calibration• Programmable• Splash proof front panel• Mounting hardware included• CE mark• Programmable set point controller• Dual channel with math functions• Digital I/O• Large LCD display• Splash proof front panel• Push-button calibration• Flame retardant mini-module housing• Master/slave excitation synchronization(Up to 10 channels)• LED status lights• Supports all standard AC LVDTs, RVDTs,
and VR half-bridge sensors

Dimensions (mm)173 x 97 x 49

178 x 92 x 92

85.1 x 70.4 x 17.8

Typical Applications

Remote monitoring stations, measurement
test stands, process monitoring

LVDT based weighing systems, pass/
fail parts sorting, quality inspection

Automotive test instrumentation,
factory automation

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 37POSITION SENSORSPOSITION SENSORSLINEAR POSITION—POTENTIOMETERS

MLP, CLP
Package

Aluminum body, steel rod, IP65, IP67

Range

0 - 0.5 to 0 - 6" (MLP) 0 - 1 to 0 - 10" (CLP)

Linearity±0.5 to ±1% (MLP) ±0.1 to ±0.2% (CLP)

Excitation

Up to 40 VDC max.

OutputVoltage divider

Resolution

Essentially infinite

Max. Speed

10 m/sUnique Features• Extended temperature range, miniature design• First choice for auto racing applications• Perfect for high cycle applications

Operating Temp.

-40°C to 90°C

Dimensions (mm)Diameter/cross section: Ø9.5 mm (MLP) 15 mm x 15 mm (CLP)

Typical Applications

Vehicle testing, autosport instrumentation, structuraland architectural testing and robotics.

LINEAR POSITION—POTENTIOMETERS

5903, 5905

Linear Motion
Package• 7.94 mm - 12.7 mm/0.312" - 0.500" housing diameter• 1.98 mm - 3.18 mm/0.078" - 0.125" shaft diameter

Resistance

1K/5K/10K

Range

5903 series - up to 50.8 mm/2" stroke
5905 series - up to 101.6 mm/4" stroke

Linearity±1%

Output Smoothness<0.1%

Resolution

Infinite

Operating Temp.

-65°C to 125°C

Stroke Life

50 million cycles min

Typical Applications

Critical position feedback applications in commercial,industrial, medical, aircraft and military markets

PAGE 38POSITION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

POSITION SENSORSANGULAR POSITION—POTENTIOMETERS

6000

6200

6900

6100

Package• 12.7 mm - 50.8 mm/0.500"- 2.00" housing diameter• 3.170 mm - 6.34 mm/0.1248"- 0.2498" shaft diameter• 12.7 mm - 1.74 mm/0.500"- 0.680" housing length• 11.11 mm - 47.62 mm/0.438"- 1.875" mounting
pilot diameter• 12.7 mm - 50.8 mm/0.500"- 2.00" housing diameter• 3.170 mm - 6.34 mm/0.1248"- .2498" shaft diameter• 12.7 mm - 1.74 mm/0.500"- 0.680" housing length• 3/8 32 NEF thread/10.31 mm/
0.4062" pilot diameter• 17.81 mm - 45.85 mm/0.702" - 1.805
in element outside diameter• 4.724 mm - 11.05 mm/0.186" 0.435" element inside diameter• 3.175 mm -6.35 mm/0.125" - 0.250
shaft insulator inside diameter• 4.064 mm - 7.80 mm/0.160" - 0.307"
mating wiper inside diameter• 5.08 mm/0.200" assembled
package height• 27.94 mm - 66.5 mm/1.100"- 2.62" housing diameter• 3.175 mm - 19 mm/0.125" 0.752" hollow shaft diameter

Resistance

1K - 20KΩ

1K - 20KΩ

1K/5K/10KΩ

1K - 20KΩ

Range

Up to 355°

Up to 355°

Up to 350°

Up to 355°

Linearity± 0.5%± 0.5%± 0.5%± 0.5%

Output Smoothness

<0.1%

<0.1%

< 0.1%< 0.1%

Resolution

Infinite

Infinite

Infinite

Infinite

Operating Temp.

-65°C to 125°C-65°C to 125°C-65°C to 125°C-65°C to 125°C

Rotational Life

50 million cycles/minute50 million cycles/minute50 million cycles/minute50 million cycles/minute.

Typical Applications

Critical position feedback
applications in commercial,industrial, medical, aircraft
and military markets

Critical position feedback
applications in commercial,industrial, medical, aircraft
and military markets

Critical position feedback
applications in commercial,industrial, medical, aircraft
and military markets

Critical position feedback
applications in commercial,industrial, medical, aircraft
and military markets

Servo Mount

Bushing Mount

Element/Wiper/Insulator

Hollow Shaft

ANGULAR POSITION—POTENTIOMETERS

RT8, RT9
Package

Aluminum or stainless IP67, IP68

Resolution±0.15% to ±1.25%

Unique Features• Absolute rotary• Designed for heavy industrial applications• CSA, CENELEC certification for hazardous area applications

OutputVoltage divider, 0 - 5 V, 0 - 10 V, 4 - 20 mA,
incremental encoder, CANbus, DeviceNet™

Range

0 - 0.125 to 0 - 200 turns

Operating Temp.

-40°C to 90°C

Dimensions (mm)

Ø65 x 100 (RT8) Ø115 x 60 (RT9)

Typical Applications

Valve control, airport passenger loading bridge,
water management, factory automation

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 39POSITION SENSORSPRESSURE SENSORSPRESSURE
SENSORSBOARD LEVEL PRESSURE SENSORS
Digital Output and Altimeter

MS4515DO, MS4525DO

MS5803

MS5837MS5840

Package

8 pin DIL

Surface mountable

Surface mountable

Surface mountable

Type

Gage, compound (MS4515DO)
Gage, absolute, differential,
compound (MS4525DO)

Absolute

Absolute

Absolute

Pressure Range

0 - 2 to 30" H2O (MS4515DO)
0 - 1 to 150 psi (MS4525DO)

0 - 1 to 30 bar

0 - 2 bar
0 - 30 bar

0 - 2 bar
Operating range:
300 to 1200 mbar

Output/Span

14-bit ADC SPI or I2C

24-bit ADC I²C and
SPI (Mode 0, 3)24-bit ADC I2C

24-bit ADC I2C

Resolution—

12 µbar (MS5803-01BA) 0.5
mbar (MS5803-30BA)

0.016 mbar (2 bar)
0.2 mbar (30 bar)

13 cm of air

Unique Features• Optional gel coat, low power• Pressure and temperature
measurement• Single supply of 3.3 or 5.0 VDC• Top, side barbed or
manifold o-ring port
• J lead or thru hole pins

• 24-bit digital sensor, software
calibration and temperaturecompensation (I2C and SPI),
no external components• Supply voltage 1.8 to 3.6 V• Supply voltage: 1.5 to 3.6 V• Pressure and temperature
measurement• Excellent long term stability• Hermetically sealable
for outdoor devices• Sealing designed for
1.8 x 0.88 mm o-ring• Supply voltage: 1.5 to 3.6 V• Pressure and temperature
measurement• Low power, 0.6 μA(standby ≤ 0.1 μA at 25°C)• Protected against
direct sunlight

Linearity/Absolute
Accuracy

0.25%/1% TEB±1.5 mbar at 25°C (MS580301BA) ±250 mbar at 0°C to
40°C (MS5803-30BA)±.5 mbar (2 bar)±400 mbar (30 bar)

±0.5mbar at 20°C
±4mbar from -20°C to 85°C

Overpressure

300 psi

10 bar (1, 2 bar)
30 bar (5, 7, 14 bar)
50 bar (30 bar)

10 bar (2 bar)
50 bar (30 bar)—

Operating Temp.

-10°C to 85°C (MS4515DO)-25°C to 105°C (MS4525DO)-40°C to 85°C

-20°C to 85 °C-20°C to 85 °C

Dimensions (mm)12.5 x 9.9

6.4 x 6.2 x 2.9

3.3 x 3.3 x 2.75

3.3 x 3.3 x 1.7mm

Typical Applications

Medical instruments, airflow measurements, processcontrol, leak detection

Precision altimeter, diving
and multi-mode watches,in-building navigation,
variometers/flight instruments

Mobile water depth
measurement systems, diving
computers, adventure or multimode watches, data loggers

Altimeter and barometer
applications, adventure
or multi-mode watches,
drones, bike computers

PAGE 40
PRESSURE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

PRESSURE SENSORSBOARD LEVEL PRESSURE SENSORS
Digital Output and Altimeter

MS5525DSO

MS5607, MS5611, MS5637

Package

SOIC-14

Surface mountable

Type

Gage, absolute, differential, compound

Absolute

Pressure Range

0 - 1 to 30 psi

10 - 2K mbar

Output/Span

24-bit ADC SPI or I2C protocol

24-bit ADC I²C

Resolution—

0.016 mbar

Unique Features• 24-bit digital small outline sensor• Pressure and temperature measurement• Single supply of 1.8 or 3.6 VDC• Barb, tube and hole package style options• 24-bit digital sensor• 13 cm resolution (MS5607, MS5637)• 10 cm resolution (MS5611)• Supply voltage: 1.5 to 3.6 V (MS5637) Supply voltage: 1.8 to 3.6 V(MS5607, MS5611)• Low power, 0.6 µA (Standby ≤ 0.1 µA at 25°C)

Linearity/Absolute
Accuracy

0.25%/2.5% TEB±2.0 mbar at 25°C

Overpressure

3X range

6 barOperating Temp.

-40°C to 125°C-40°C to 85°C

Dimensions (mm)12.5 x 7.93 x 3 x 0.9 (MS5637) 5 x 3 x 1 (MS5607, MS5611)

TypicalApplications

Medical respirators, ventilators, factory automation, altitude andairspeed measurements, leak detection, home appliances

Smart phones, tablets, personal navigation devices,tire pressure monitoring, compressors

MS5805MS8607

Package

Surface mountable

Surface mountable

Type

Absolute

Absolute

Pressure Range

10 - 2K mbar

10 - 2K mbar

Output/Span

24-bit ADC I²C

24-bit ADC I²C

Resolution

0.02 mbar

0.016 mbar

Unique Features• 24-bit digital sensor• 20 cm resolution• Supply voltage: 1.8 to 3.6 V• Sealing designed for 2.5 x 1 mm o-ring• Silicone gel protection• Waterproof• Integrated pressure, humidity and temperature• Supply voltage: 1.5 to 3.6 V• Fully factory calibrated sensor

Linearity/Absolute
Accuracy±2.0 mbar at 25°C

±4 mbar

Overpressure

5 bar

6 bar

Operating Temp.

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)4.5 x 4.5 x 3.5

5x3x1

TypicalApplications

Mobile altimeter and barometer systems, bike computers,adventure or multi-mode watches, variometers, data loggers

Mobile water depth measurement systems, diving computers,adventure or multi-mode watches, data loggers

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 41
PRESSURE SENSORSPRESSURE SENSORSBOARD LEVEL PRESSURE SENSORSmV Output

MS1451, MS1471

MS52xx, MS54xx

Package

Surface mountable

Surface mountable

Type

Gage, absoluteGage, absolute

Pressure Range

0 - 5 to 500 psi

0 - 1 to 12 bar

Output/Span

60 mV typical

150 mV, 240 mV

Unique Features• Low cost
• Coarse calibrated at room temp. (MS1471)• With gel to protect against moisture• Tube or hole

• Small size (MS54xx)• High linearity or high sensitivity options• Plastic tube or metal ring options• With gel to protect against moisture• High endurance (Option HM)

Accuracy±0.25% non-linearity±0.05%, ±0.15% FS non-linearity (MS52xx)±0.05%, ±0.2% FS non-linearity (MS54xx)

Operating Temp.

-40°C to 125°C-40°C to 125°C

Dimensions (mm)7.6 x 7.6, application dependent

7.6 x 7.6, application dependent (MS52xx) 6.4 x 6.2 (MS54xx)

Typical Applications

Altitude measurement, barometric pressure, medical
instrumentation, consumer appliances, tire pressure

Absolute pressure sensor systems, engine controls, high resolution
altimeters, variometers, waterproof watches, diver computers,
barometers, tire pressure monitoring systems (TPMS), medical
instrumentation, pneumatic controls

DISPOSABLE MEDICAL PRESSURE SENSORSmV Output

1620, 1630

Fully Assembled 1620

Package

Hybrid assembly

Plastic housing

Type

Gage

Gage

Pressure Range-30 to 300 mmHg

-30 to 300 mmHg

Output/Span

5 µV/V/mmHg

5 µV/V/mmHg

Unique Features• Low cost, disposable design• Supplied in tape and reel
• Compliant to AAMI spec
• ISO13485 certified• Low cost, disposable design• Compliant to AAMI spec
• Custom designs available

Operating Temp.

10°C to 40°C

10°C to 40°C

Dimensions (mm)1620: 11.43 x 8.13 x 4.20 1630: 12.7 x 5.08 x 3.94

42.8 x 30.3 x 19.0

Typical Applications

Disposable blood pressure, surgical procedures, ICU,
kidney dialysis machines, medical instrumentation

Disposable blood pressure, kidney dialysis machines,
surgical procedures and intensive care units. Ready touse, fully assembled disposable sensor units with cable,
connector, stop cock, flush device in a plastic housing.

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

(Customized per customer specifications)te.com/sensors

PAGE 42PRESSURE SENSORSPRESSURE SENSORSMEDIA ISOLATED PRESSURE SENSOR MODULES
Digital Output

85BSD

86BSD

89BSD

154BSD

Package• 13 mm diaphragm diameter• Weldable or threaded
process fittings• 16 mm diaphragm diameter• O-ring mount• 9 mm diaphragm diameter• Threaded or weldable• 19 mm diaphragm diameter• O-ring mountType

Gage, absoluteGage, absolute

Absolute, sealed gage

Gage, absolute

Pressure Range

0 - 0.35 to 20 bar /0 - 5 to 300 psi

0 - 0.07 to 20 bar /0 - 1 to 300 psi

0 - 6 to 30 bar

0 - 1 to 300 psi

Output/Span

14-bit ADC I2C or SPI

14-bit ADC I2C or SPI

24-bit ADC I2C

14-bit ADC I2C or SPI

Unique Features• Pressure and temperature
read-out• Cable and connector options• Low power option• Pressure and temperature
read-out• Cable and connector options• Low power option• Pressure and temperature
read-out• Low power: 1 µA
(Standby < 0.15 µA)• Pressure and temperature
read-out• Cable and connector options• Low power option

Accuracy±0.25% span

±0.25% span

±0.3% span

±0.25% span

Total Error Band

±1.0% FSO

±1.0% FSO

±3.0% FSO max.

±1.0% FSO

Overpressure

2X

2X

2X

2X

Operating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 85°C

-40°C to 125°C

Dimensions (mm)Ø15.85 x 7.9

Ø15.82 x 9.3

Ø9.04 x 7.5

Ø19 x 13.8

Typical Applications

Level controls, tank levelmeasurement, corrosive fluids
and gas measurement systems,
sealed systems, manifoldpressure measurement,
submersible depth monitoring

Level controls, tank levelmeasurement, corrosive fluids
and gas measurement systems,
sealed systems, manifoldpressure measurement,
submersible depth monitoring

Level controls, tank levelmeasurement, corrosive fluids
and gas measurement systems,
sealed systems, manifold pressure
measurement, dive computers

Level controls, tank levelmeasurement, corrosive fluids
and gas measurement systems,
sealed systems, manifoldpressure measurement,
submersible depth monitoring

TRANSDUCERS AND TRANSMITTERS
Industrial Wireless

M5600, U5600

MSP100MSP300, MSP340

Type

Gage, sealed, absolute, compound

Gage

Gage

Pressure Range

0 – 50 to 15K psi (M5600),
0 – 5 to 10K psi (U5600)

0 - 100 to 500 psi

0 - 100 to 10K psi (MSP300)
0 - 50 to 10K psi (MSP340)

Output/Span

24-bit ADC I2C

100 mV typical0 - 100 mV, 0.5 - 4.5 VDC,
1 - 5 VDC, 4 - 20 mA

Unique Features• Pressure and temperature• 2.3 – 3.6 V supply voltage• Compact and battery-powered• Weather resistant (IP66 and IP67)• Stainless steel and polycarbonate enclosure• Microfused• Low cost stainless steel isolated transducer• No threads needed for pressure connect• Highly customized for OEM application• Small size
• Solid state reliability• Microfused• Highly customized for OEM applications• Small size
• Solid state reliability

Accuracy±0.25% FS (M5600)
Down to ±0.1% FS (U5600)±0.5% FSO

±1% FSO

Operating Temp.

-20°C to 85°C

0°C to 55°C

-20°C to 85°C

Dimensions (mm)24 x 24 x 69

12.7 x 24.38 x 20.32

MSP300: 22.23 x 22.23 x 55.88
MSP340: 15.88 x 15.88 x 75.44

Typical Applications

Industrial process control and monitoring,
advanced HVACR systems, refrigeration
systems, automotive test stands, off-roadvehicles, pumps and compressors, hydraulicand pneumatic systems, agriculture equipment,
energy generation and management

Beverage dispensing systems, automation,
HVACR controls, energy and water
management, pumps, compressors,pneumatic equipment

Paint sprayers, braking systems, HVACR
controls, energy and water
management, pumps, compressors,pneumatic equipment, off road heavyequipment, agriculture equipment

Agency Approvals

CE, FCC—UL 508 (MSP300)

PAGE 43
PRESSURE
SENSORSte.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

PRESSURE SENSORSMEDIA ISOLATED PRESSURE SENSOR MODULES
Analog Output

82, 85 with Fittings

89 Button, 89 with Fittings

86A Amplified

Package

Weldable (85) or process fitting

Weldable or process fitting

5/8” (16 mm) diameter o-ring mountType

Gage, absolute, vacuum gage

Sealed gage, absoluteGage, absolute

Pressure Range

0 - 5 to 500 psi (85), 0 - 1 to 500 psi (82)

0 - 1K to 10K psi

0 - 1 to 150 psi

Output/Span

100 mV typical

100 mV typical

0.5 - 4.5 VDC

Unique Features• Modular design• High pressure• Modular design• Small diameter, amplified output• Bar ranges available

Non-linearity±0.3% FSO (1 psi)
±0.2% FSO (5 psi)
±0.1% FSO (≥15 psi)

±0.25% FSO±1.0% FSO

Operating Temp.

-40°C to 125°C-40°C to 125°C-20°C to 85°C

Dimensions (mm)Fittings: application dependent

89 Button: Ø9.04 x 13.2
89 with Fittings: application dependentØ15.82 x 9.3

Typical Applications

Medical, process control, refrigeration
compressor, oceanography, level systems

Air tank pressure, hydraulics, process control,
robotics, refrigeration compressors,
oceanography

Level measurement, OEM transmittersand transducers, process control

MEDIA ISOLATED PRESSURE SENSOR MODULES
Analog Output

82, 85, 85F, 86, 154N

DP86 O-Ring Mount,with Fittings/Cable

U86B

Package• 3/4" (19 mm) diameter o-ring
mount (82, 154N)• 5/8" (16 mm) diameter o-ring mount (86)• 1/2" (13 mm) diameter o-ringflush mount (85F)• 1/2" (13 mm) diameter o-ring mount (85)• 5/8" (16 mm) diameter o-ring
mount or threaded process fittings• Mountable with o-ring sealType

Gage, absolute, vacuum gage (82, 85,
86, 154N) Gage, absolute (85F)

Differential

Sealed gage, absolute

Pressure Range

0 - 1 to 500 psi (Absolute, gage: 82, 154N)
0 - 5 to 500 psi (Absolute, gage: 85, 86)
0 - 15 to 500 psi (85F, vacuum gage:82, 85, 86, 154N)

0 - 1 to 500 psi

0 - 5 to 13 bar/0 - 50 to 200 psi

Output/Span

100 mV typical

100 mV typical/sensitivity dependent

0.5 - 4.5 VDC(Ratiometric output)

Unique Features• High performance• High stability for OEM applications• Minimizes trapped volume (85F)• Wet/wet differential pressure• Line pressure max. 1,000 psi• Amplified

Non-linearity±0.3% FSO (1 psi), ±0.2% FSO (5 psi)
±0.1% FSO (≥15 psi), ±0.1% FSO (85F)±0.3% FSO (1 psi)
±0.2% FSO (5 psi)
±0.1% FSO (≥15 psi)

±0.5% FSO

Operating Temp.

-40°C to 125°C (82 / 85 / 86 /154N), -20°C to 125°C (85F)-40°C to 125°C-7°C to 105°C

Dimensions (mm)82: Ø19 x 6.48
154N: Ø18.97 x 13.8
85: Ø15.85 x 9.3

O-ring: Ø15.82 x 17.5
Fittings: Application dependentØ15.82 x 13.6
Socket spacing: 31.75

TypicalApplications

Hydraulic controls, process control,
oceanography, refrigeration/compressors,pressure transmitters, level systems, dialysis
machines, infusion pumps, medical systems

Level controls, tank level measurement,
corrosive fluids and gas measurement
systems, flow measurement

Urea level, urea pressure, air brakes,
corrosive fluid measurement forengine & vehicle applications

SS-TS-TE101
06/201986: Ø15.82 x 11.4
85F: Ø17.2 x 11.33

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 44PRESSURE SENSORSPRESSURE SENSORSTRANSDUCERS AND TRANSMITTERS
Industrial

US300

AST20HA, AST20PT, AST20SW

AST4000, AST4100, AST4200

Type

Gage, absoluteGage, sealed gage, absoluteGage, sealed gage, compound

Pressure Range

0 - 15 to 5K psi

0 - 1 to 60K psi

0 - 25 to 10K psi

Output/Span

0 - 10 mV/V, 0.5 - 4.5 V, 1 - 5 V, 4 - 20 mA

0.5 - 4.5 V (Ratiometric)
1 - 5 V 4 - 20 mA, 0 - 5 V, 0 - 10 V,
switch (AST20SW)

0.5 - 4.5 V (Ratiometric), 1 - 5 V, 1 - 10 V,
4 - 20 mA, 0.5 - 2.5 V

Unique Features• UltraStable technology• Highly customized for OEM applications• Small size
• Solid state reliability• Excellent performance over temperature• Semi-custom designs available• Fault mode condition settings• Four standard sensor material options• Additional temperature output (AST20PT)• Four standard sensor material options• Rugged construction• 100 V/m EMI/RFI protection• Semi-custom designs available

Accuracy±0.15% FSO (15 - 1K psi), ±0.25% FSO (>1K psi)

±0.1% FSO

±0.5% FSO

Operating Temp.

-40°C to 105°C-40°C to 85°C

-40°C to 85°C

Dimensions (mm)15.88 x 15.88 x 98.00

Application dependent

Application dependent

Typical Applications

Paint sprayers, braking systems,
HVACR controls, energy and water
management, pumps, compressors,pneumatic equipment, off road heavyequipment, agriculture equipment

Test and measurement, industrial controls

Water, hydraulic equipment, HVACR,
industrial controls

Agency Approvals—

ABS, CE

UL/cUL508, ABS, CE

M3200

M5200

U5200, U5300

D5100

Type

Gage, CompoundGage, sealed, compoundGage, sealed, absolute, compound

Differential wet/wet

Pressure Range

0-7bar to 500 bar /0-100 to 7,500 Psi

0 - 3.5 to 1K bar/0 - 50 to 15K psi

0 - 0.14 to 700 bar/0 - 2 to 10K psi

0 - 0.07 to 35 bar/0- 1 to 500 psi

Output/Span

100mV, 0.5 - 4.5 V, 1 - 5 V,
0 - 5 V, 0 - 10 V,
4 - 20 mA and I2C Digital Output

0.5 - 4.5 V, 1 - 5 V,
0 - 5 V, 0 - 10 V,
4 - 20 mA, 1 - 6 V

0.5 - 4.5 V, 1 - 5 V,
0 - 5 V, 0 - 10 V,
4 - 20 mA, 1 - 6 V

80 mV / 100 mV, 0.5 - 4.5 VDC,
1 - 5 VDC, 4 - 20 mA

Unique Features• Microfused technology• High performance at a low cost
• Solid state reliability• ±1.5% FSO TEB (-20°C to 85°C)• Weatherproof - IP67 with cable• 14bit digital output for Pressure• Microfused technology• High performance at a low cost
• Solid state reliability• ±1% FSO TEB (-20°C to 85°C)• Weatherproof• 17 - 4 PH or 316L SS• UltraStable technology• High performance at a low cost
• ±0.75% FSO TEB (-20°C to 85°C,>5 psi and ≤5000 psi) (U5200)• ±0.5% FSO TEB (-20°C
to 85°C) (U5300)• Weatherproof• High accuracy (U5300)• UltraStable technology• High performanceat a low cost
• Solid state reliability• ±1% FSO TEB (-20°C to 85°C)• Line pressure max. 1000 psi

Accuracy±0.25% FSO±0.25% FSO±0.1% FSO (>5 and ≤500 psi)

±0.3% FSO (<5 psi), ±0.25%
FSO (5 psi), ±0.1 % FSO (≥15 psi)

Operating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 125°C-40°C to 125°C

Dimensions (mm)22.2 on Hex x 58 max

24 X 24 X 82 max.

24 X 24 X 82 max.

25.4 x 58.4 x 72.0

Typical Applications

Suitable for measurement ofliquid or gas pressure, evenfor difficult media such as
contaminated water, steam,and mildly corrosive fluids.

Industrial process control andmonitoring, advanced HVACR
systems, refrigeration systems,
automotive test stands, off roadvehicles, pumps and compressors,
hydraulic and pneumatic systems,
agriculture equipment, energy
generation and management

Industrial process control andmonitoring, advanced HVACR
systems, refrigeration systems,
automotive test stands, off roadvehicles, pumps and compressors,
hydraulic and pneumatic systems,
agriculture equipment, energy
generation and management,
military and aerospace test
stands, calibration equipment,high accuracy applications,
stationary motor fuel control,high end industry machinery

Process controls, tanklevel measurement, filter
performance monitoring,corrosive fluids and gas
measurement systems,
flow measurement

Agency Approvals

CE (EMC)CE (EMC)CE (EMC), UL 508

CE (EMC)PAGE 45
PRESSURE
SENSORSte.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

PRESSURE SENSORSTRANSDUCERS AND TRANSMITTERS
Heavy Industrial

M7100, U7100

P900, P981, P1200, P700, P9000

P101, P105, P125

Type

Gage, no vent gage (M7100) Gage,
sealed gage, absolute (U7100)

Gage, absolute

Gage

Pressure Range

0 - 10 to 700 bar/0 - 150 to 10K psi (M7100)
0 - 1 to 10 bar/0 - 15 to 150 psi (U7100)

0 - 5 bar to 700 bar/0 - 75 to 10K psi

0 - 10 to 7K bar/0 - 150 to 100K psi

Output/Span

0.5 - 4.5 VDC [Ratiometric output] 1 - 5
VDC [Regulated] (M7100) 0.5 - 4.5 VDC[Ratiometric output] (U7100)

0 - 5 VDC, 0 - 10 VDC, 4 - 20 mA

7.5 to 20 mV(4 V; 5 V optional)

Unique Features• ±1% FSO TEB (-20°C to 85°C)• Solid state reliability• Survives high vibration and immersion• Microfused technology (M7100)• UltraStable technology (U7100)• Copper tube for HVACR (M7100)• High overpressure (10X over pressure)• Shock and vibration resistant• Heavy industrial grade transducer (P9000)• Advanced digital compensation / calibration• Mechanical over pressure stops• High temperature operation• Stainless steel diaphragm• Female pressure connectors:
M16 x 1.5, M20 x 1.5, 1/4 NPT• Metal to metal seal

Accuracy

0.25% FSO

0.1% to 0.2% FSO±0.3% FSO

Operating Temp.

-40°C to 125°C-54°C to 120°C-20°C to 80°C

Dimensions (mm)26.7 x 26.7 x 50.0

Application dependentØ29 x 85 max.

Typical Applications

HVACR refrigeration controls, off roadvehicles engine control, compressors,
hydraulic, energy and water management

Steel mills, hydraulic controls, power generationequipment, torpedo depth, military andaerospace, vehicle braking systems

Harsh environments, aggressive liquids

Agency Approvals

CE (EMC), UL 508­

CE, CENELEC (Intrinsically Safe)—

TRANSDUCERS AND TRANSMITTERS
Miniature

XP

XPC10

Type

Gage, sealed, absoluteGage, sealed, absolute

Pressure Range

0 - 1 to 350 bar/0 - 15 to 5K psi (XP5, XPM10)
0 - 5 to 200 bar/0 - 75 to 3K psi (XPM4)
0 - 100 to 1K bar/0 - 1.5K to 15K psi (XPM6)

0 - 10 to 500 bar/0 - 150 to 7.5K psi

Output/Span

20 - 100 mV, 4 V FSO (Amplified)

12 mV FSO, 4 V FSO (Amplified)

Unique Features• Titanium construction (XP5, XPM4)• Stainless steel housing (XPM6, XPM10)• Amplified output options (XP5, XPM6, XPM10)• Cable and connector options• For static and dynamic applications• Amplified output available• For static and dynamic applications• Optional IP67 ingress protection• High temperature operation

Accuracy

Down to ±0.25% FSO (XP5, XPM6, XPM10), down to ±0.35% FSO (XPM4)

Down to ±0.25% FSO

Operating Temp.

-40°C to 120°C-40°C to 220°C

Dimensions (mm)

XPM4: M4 x 0.7 thread; Hex 8
XP5: M5 x 0.8 or 10-32 UNF thread; Hex 10
XPM6: M6 x 1 thread; Hex 12
XPM10: M10 x 1 thread; Hex 15

M10 x 1 or 3/8-24 UNF thread; Hex 15

Typical Applications

Corrosive liquids and gases, braking system pressure, onboard equipment
monitoring, military and aerospace, explosive test benches, robotics
and effectors, laboratory and research, extreme miniature devices

Aerospace, test benches, oven monitoring equipment,cooling regulation systems

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 46
PRESSURE SENSORSPRESSURE SENSORSTRANSDUCERS AND TRANSMITTERS
Miniature

EB, EPRB

EPIH

EPB, EPB-PW, EPL

Type

Gage, sealed, absoluteGage, sealed, absoluteGage, sealed, absolute

Pressure Range

0 - 0.35 to 700 bar/0 - 5 to 10K psi

0 - 0.35 to 20 bar/0 - 5 to 300 psi

0 - 0.35 to 350 bar/0 - 5 to 5K psi

Output/Span

0.5 to 4.5 VDC

12 mV to 75 mV

10 mV to 125 mV

Unique Features• High accuracy• Miniature design• UltraStable technology• EMI protected• Combined pressure and temperature• Diffused silicon diaphragm with a large
variety of sizes and shapes available
as small as 0.05” outside diameter• High frequency response (To 1.7 MHz)• Ultra-miniature design• Miniature flush mountable• Flush stainless steel diaphragm,
flanged or non-flanged• Bonded silicon gage, high frequency
response (To 400 KHz)
• IP68 ingress protection in Titanium
construction (EPB-PW)

Accuracy±0.25% FSO±1.0% FSO

±0.5 to ±1% FSO

Operating Temp.

-40°C to 125°C (Available option up to 150°C)-40°C to 120°C-40°C to 120°C

Dimensions (mm)11 body diameter

Application dependent

3.2 to 7 outside diameter

Typical Applications

Motor sport, hydraulic/pneumatic
systems, automotive test stands,
military and aerospace test stands

Aerospace testing, wind tunnels, biomedical
testing, aircraft body and wing dynamics,
high frequency measurements

Air flow testing, hydraulic pressure
systems, air pressure systems, bearing
studies, ballistics, water hammer,
miniature scale model testing, centrifuge
pore water pressure measurements

Agency Approvals

CE (EMC)——TRANSDUCERS AND TRANSMITTERS
Liquid Level

U5700

AST45xxType

Gage, sealed, absolute, compoundGage, absolute

Pressure Range

0 - 2 to 10K psi

0 - 1 to 100 psi (AST4500, AST4510, AST4520)

Output/Span

0.5 - 4.5 V, 1 - 5 V, 0 - 5 V, 0 - 10 V, 4 - 20 mA, 1 - 6 V

0.5 - 4.5 V [Ratiometric], 1 - 5 V, 4 - 20 mA, 0.5 - 2.5 V

Unique Features• UltraStable technology• High accuracy• IP68 rated connection and submersible• Polyurethane jacketed cable• Optional Polyoxymethylene cap• Intrinsically safe ratings• Material options including: 316L, alloy C276, and PVDF
• Low power options• High quality cable options

Accuracy

0.1 % FSO

±0.25% FSO

Operating Temp.

-10°C to 60°C

-40°C to 85°C

Dimensions (mm)22.23 x 22.23 x 98.04

Application dependent

Typical Applications

Industrial process control and monitoring, advanced HVACR
systems, refrigeration systems, automotive test stands,
off road vehicles, pumps and compressors, hydraulic/pneumatic systems, agriculture equipment, energy
generation and management, liquid level applications

Diesel tanks, chemical tanks, water tanks

Agency Approvals

CE (EMC)UL/CSA Class I Div I, ATEX/IECEx Exia, ABS, CE

PAGE 47
PRESSURE
SENSORSte.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

PRESSURE SENSORSTRANSDUCERS AND TRANSMITTERS
Hazardous Location

AST43xx, AST44xx

AST46xxType

Gage, sealed gage, compound, absoluteGage, sealed gage, compound, absolute

Pressure Range

0 - 1 to 15 psi (AST43LP, AST44LP) 0 - 25 to 20K psi(AST4300, AST4400, AST4401)

0 - 1 to 20K psi

Output/Span

0.5 - 4.5 V [Ratiometric], 1 - 5 V, 4 - 20 mA, 0.5 - 2.5 V

0.5 - 4.5 V [Ratiometric], 1 - 5 V, 4 - 20 mA,
0.5 - 2.5 V, switch (AST46SW)

Unique Features• Available with 316L, Hastalloy C276, or Inconel 718 materials• Low current consumption options• Low power options• High proof and burst pressure• Available with 316L, Hastalloy C276, or Inconel 718 materials• Low current consumption options• Low power options• Local display (AST46DS)• Additional temperature output

Accuracy±0.25% FSO±0.25% FSO (AST4600, AST46DS), ±0.1% FSO (AST46HA, AST46PT)

Operating Temp.

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)Application dependent

Application dependent

Typical Applications

Compressors, well sites, ships, factory automation,
SCADA equipment, offshore equipment

SCADA/RTU, well sites, offshore equipment, hydraulic controls

Agency Approvals

UL/CSA Class I Div I and II, ATEX/IECEx Exia/
Exn, CCOE, CNEx, ABS, CE

CSA Class I/II Div I, ATEX/IECEx Exd, ABS, CE

AST5100, AST5300, AST5400

AST2000H2

Type

Differential

Gage, sealed gage

Pressure Range

0 - 5 H2O to 5K psi

0-10 PSI to 10000 PSI

Output/Span

0.5 - 4.5 V [Ratiometric], 0 - 5 V, 1 - 5 V, 4 - 20 mA

0.5 - 4.5 V [Ratiometric] 1 - 5 V, 4 - 20 mA

Unique Features• Wide range of pressures available• Full line pressure on either side without zero shifts• Hazardous location approvals (AST5300, AST5400)• 20 bar, 448 bar, 900 bar• High pressure H2 Storage• CE EN61326

Accuracy±0.25% FSO (AST5100, AST5300), 1% TEB (AST5400)±0.25% BFSL

Operating Temp.

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)Application dependent

Application dependent

Typical Applications

Filter monitoring, flow measurement, tank level measurement

PEM fuel cells, hydrogen storage, hydrogen
filling stations, backup power

Agency Approvals

CSA Class I/II Div I and II, ATEX/IECEx Exd/Exn, ABS, CE

EC-79 e24*79/2009*406/2010*0006*02 CE EN61326

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 48
PRESSURE SENSORSRATE AND INERTIAL SENSORSRATE ANDINERTIALSENSORSRATE SENSORS AND GYROS

Package

11206AC

11207AC

31206B/31207B

610

603

Electroless nickel
plated Aluminum

Electroless nickel
plated Aluminum

Electroless nickel
plated Aluminum

Anodized aluminum

Anodized aluminum

FS Ranges±50, ±180°/sec±300°/sec±50, ±180, ±1,000°/sec±500 to ±50K°/sec±100 to ±24K°/sec

UniqueFeatures• IdentiCal™
interchangeable sensor• ±0.5% accuracy from
-40°C to +85°C• Silicon MEMS gyro
• EN61000-6-2/-4 certified
for industrial environment• IdentiCal™
interchangeable sensor• High stability• Low noise• Vibration-rejecting• Triaxial angular rate sensor• Stable performanceover temperature• Power supply regulation• Temperature
calibration data• Small, lightweight package• SAE-J211, ISO-6487,
NHTSA approved• Crash testing certified• Insensitive to shock• MEMS triaxial rate sensor• SAEJ211 compliant• Shock resistant housing• Rugged compact package

Accuracy±0.1% non-linearity±0.01% non-linearity±0.1% non-linearity±0.5% non-linearity±0.5% non-linearity

Excitation
Voltage

8.5 - 36 VDC

10 - 36 VDC

8.5 - 36 VDC

5 - 16 VDC

5 - 16 VDC

Operating
Temp.

-40°C to 85°C

-40°C to 85°C

-40°C to 85°C

-40°C to 105°C-40°C to 105°C

Dimensions(mm)

24 x 24 x 27.30

24 x 24 x 27.30

24 x 24 x 28.30

14.6 x 10.2 x 7.6

20.8 x 20.8 x 14.5

TypicalApplications

Wind turbine, weapons
testing, test andmeasurement

Wind turbine, weapons
testing, test andmeasurement

Weapons testing,boat stabilization,
test and measurement

Automotive safety crashtesting, roll-over testing,motor sports, biomechanics,
weapons testing

Automotive safety crashtesting, pedestrian impact,biomechanics, robotics

PAGE 49
RATE AND INERTIAL SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

RATE AND INERTIAL SENSORS
6 DEGREES OF FREEDOM SENSORS

633

634

Package

Stainless steel

Anodized aluminum

FS Ranges±500 to ±50K°/sec±50g to ±6,000g±100 to ±18K°/sec±2g to ±100g

Unique Features• Complete six degree-of-freedom (6DoF) analog sensor• Shock resistant rugged housing• Silicon MEMS gyros• PR MEMS high-g shock sensors• 6DoF analog sensor• Signal conditioned output• Silicon MEMS gyros• UltraStable VC MEMS low-g sensors

Accuracy±0.5% non-linearity±0.1% non-linearity

Excitation Voltage

5 - 16 VDC

5 - 16 VDC

Operating Temp.

-40°C to 105°C-40°C to 105°C

Dimensions (mm)21.3 x 21.3 x 15.2

30.5 x 30.5 x 24.6

Typical Applications

Aerospace testing, weapons testing, biomechanics, shock andimpact testing

Automotive testing, motion measurements, biomechanics

INERTIAL SENSORS

65210E

65210ES

Package

Electroless nickel plated Aluminum

Electroless nickel plated Aluminum

FS Ranges±1K to ±20K°/sec, ±10g to ±100g±1K to ±20K°/sec, ±10g to ±100g

Unique Features• 6DoF and telemetry kit• User configurable, IRIG encoder• Parabolic and radome ring mounting adaptor options• Up to 4hr battery options• 6DoF and telemetry kit• Parabolic and radome ring mounting adaptor options• AES encryption option• Wide range of signal conditioning modules

Accuracy±0.2% non-linearity error±0.2% non-linearity error

Excitation Voltage

Li-Ion battery included

Li-Ion battery included

Operating Temp.

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)Ø69.85 x 197.3 lengthØ69.85 x 161.3 length

Typical Applications

Weapons separation testing, captive carry testing,
GTV and JTV test vehicles

Weapons separation testing, captive carry testing,
GTV and JTV test vehicles

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 50
RATE AND INERTIAL SENSORSSCANNERS AND SYSTEMSSCANNERSAND SYSTEMSPRESSURE AND TEMPERATURE SCANNERS
NetScanners

NetScanner 9216

NetScanner 9146-R

NetScanner 9146-T

NetScanner 9022

Measurement Type

Pressure

Temperature

Temperature

Pressure

Media

Dry

RTD/TC/VoltTC

LiquidAccuracy±0.05% FS±0.25°C±0.25°C±0.05% FS# of Channels

16

16/32

16

12

EU Throughput Rate

500 Hz/chan/sec

33 Hz/chan/sec

33 Hz/chan/sec

100 Hz/chan/sec

Operating Temp.

-30°C to 80°C

-30°C to 70°C

-30°C to 70°C

-30°C to 70°C

Enclosure

IP66/30 g vibration

IP66/30 g vibration

IP54/30 g vibration

IP64/30 g vibration

Typical Applications

Engine testing, portabledata acquisition, wind tunnel
research, process monitoring

Engine testing, portabledata acquisition, wind tunnel
research, process monitoring

Engine testing, portabledata acquisition, wind tunnel
research, process monitoring

Engine testing, third party
transducers, close coupled
requirements, high pressure

PAGE 51SCANNERS AND SYSTEMS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

SCANNERS AND SYSTEMSPRESSURE SCANNERS & TRANSDUCERSNetScanners, Transducers, and Accompanying Equipment

NetScanner 9032, 9033

NetScanner 9034, 9038

NetScanner 9916,
98RK-1 Rack

NetScanner 9400
Transducer

Measurement Type

Barometer, Differential Standard

Calibrator

Pressure

Pressure

Media

Dry

Dry

Dry

All-media

Accuracy±0.01% FS±0.01% FS±0.05% FS±0.05%# of Channels

1

1

128

1

EU Throughput Rate

10 Hz

10 Hz

100 Hz/chan/sec

Analog Output

Operating Temp.

-10°C to 60°C

-10°C to 60°C

0°C to 50°C

-30°C to 100°C

Enclosure

Laboratory grade

Laboratory grade

19" rackmount/4U

IP66

Typical Applications

Barometric monitor,
precision reference

Calibration, transfer standard,
verification testing

Turbine engine test, control
room location

Turbine Engine Test,
liquids, hydraulics

ELECTRONIC & MINIATURE PRESSURE SCANNERS
ESP & MicroScanner

ESP 64HD DTC

ESP 32HD DTC

ESP 64HD, 32HD, 16HD

MicroScanner 16MS

Type

Pressure

Pressure

Pressure

Pressure

Media

Dry

Dry

Dry

Dry

Accuracy±0.03% FS±0.03% FS±0.05% FS±0.05%# of Channels

64

32

64, 32 or 16

16

Thermal Comp.

Active (DTC)

Active (DTC)

Passive

ActiveOperating Temp.

-25°C to 80°C

-25°C to 80°C

-25°C to 80°C

-20°C to 100°C

Multiplex Scan Rate

70 KHz70 KHz20 KHz100 KHzPort Sizes (Inches)

0.040

0.040 or 0.063

0.040 or 0.63

Direct mountTypical Applications

Wind tunnel research, flight
test, on vehicle researchWind tunnel research, flight
test, on vehicle researchWind tunnel research, flight
test, on vehicle research

For confined space, windtunnel, flight test

DATA ACQUISITION SYSTEMSMulti-Scanner Data Acquisition Systems

Optimus

Initium

mSDI Interface

Pneumatics

Type

Pressure scanning

Pressure scanningA/D conversion

Quick disconnect

Media

Dry

Dry

Dry

Dry

Accuracy±0.03% FS±0.05% FS±0.05% FS—# of Channels

2048

512

512

19, 31, 36, 55

EU Throughput Rate

2000 Hz

1200 Hz2000 Hz—Operating Temp.

0°C to 50°C

0°C to 70°C

-25°C to 80°C

-40°C to 80°C

Enclosure

Laboratory grade

Laboratory grade

Miniature

Miniature

Typical Applications

Aerospace development

Wind engineering

In-model placement,
Optimus System interface

Pressure connections forconfined spaces

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 52
SCANNERS AND SYSTEMSSPEED SENSORSSPEEDSENSORS

PAGE 53
SPEED SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

SPEED SENSORS
3 & 4 CHANNELS HALL EFFECT
SPEED SENSORSDUAL CHANNEL HALL EFFECT
SPEED SENSOR

Jaquet DSD 25
Technology

Differential Hall Effect threeand four channels

Package• Stainless steel• Shaft lenght 29 mm• Shaft diameter 24.5 mm

Frequency Range

0 - 20 kHz

Nominal Supply Voltage

Nominal 15VDC (9 VDC to 30 VDC)

Output Signal

3 & 4 channels push-pull

Operating Temp.

-40°C to 125°C

Typical Applications

Railway

Jaquet DSD 70
Technology

Differential Hall Effect two channels

Package• Stainless steel• Shaft lenght 29 mm• Shaft diameter 16 mm

Frequency Range

0 - 20 kHz

Nominal Supply Voltage

Nominal 15VDC (9 VDC to 30 VDC)

Output Signal

2 channels push-pull

Operating Temp.

-40°C to 125°C

Typical Applications

Railway

EDDY CURRENT SPEED SENSORS

Jaquet DSH

Jaquet DSH 16

Technology

Eddy Current single channel

Eddy Current two channels

Package• Stainless steel• Shaft lenght 42 mm• Shaft diameter 18 mm• Stainless steel• Shaft lenght 29 mm• Shaft diameter 16 mm

Frequency Range

up to 20 kHz

up to 20 kHz

Nominal Supply Voltage

10 - 30VDC

Nominal 15 VDC (8 VDC to 30 VDC)

Output Signal

Square Wave

2 channels push-pull

Operating Temp.

-40°C to 125°C

120°C

Typical Applications

Industrial (Power Generation, Hydraulic, Engines, Industry)

Railway

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 54
SPEED SENSORSSPEED SENSORS
HALL EFFECT SPEED SENSORS

Jaquet Green Line D

Jaquet Green Line Y12AD

Jaquet Green Line F

Technology

Differential Hall Effect single channel

Hall Speed Sensor single channel+ direction signal

Hall Effect quasi static

Package• Stainless steel• Shaft length 26 mm, 64 mm• Shaft diameter 12 mm• Stainless steel• Shaft length 36 mm• Shaft diameter 12 mm• Stainless steel• Shaft length various• Shaft diameter various

Frequency Range

5-20 kHz

0-15 kHz

0.05-15 kHz

Nominal Supply Voltage

8-32 VDC

8-32 VDC

8-25 VDC

Output Signal

Square Wave, single channel

Square Wave, single channel + direction signal

Square Wave. single channel

Operating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 125°C

Typical Applications

Industrial, non demanding,
low cost applications

Industrial, non demanding,
low cost applications

Industrial, non demanding,
low cost applications

Jaquet DSD

Jaquet DSF extended power supply

Jaquet DSF EX-ATEX

Technology

Differential Hall Effect single channel

Hall Effect

Hall Effect

Package• Stainless steel• Various shaft lengths M12,
M14, M16, M18, M22• Various shaft diameter• Stainless steel• Various shaft lengths• Various shaft diameter• Stainless steel• Various shaft lengths• Various shaft diameter

Frequency Range

up tp 20 kHz

up to 20 kHz

up to 15 kHz

Nominal Supply Voltage

8-30 VDC

8-28 VDC
10-30 VDC

9-18 VDC

Output Signal

Square Wave, single channel

Square Wave, single channel

2-wireOperating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 125°C

Typical Applications

Industrial (Power Generation, Hydraulic,Engines, Industry)

Industrial (Power Generation, Hydraulic,Engines, Industry)

Industrial (Power Generation, Hydraulic,Engines, Industry), explosion
protected, classifed areas

Jaquet DSF

Jaquet DSL

Jaquet DSS

Technology

Hall Effect

Hall Effect

Hall Effect zero speed

Package• Stainless steel• Shaft length various• Various shaft diameter• Stainless steel• Shaft length various• Various shaft diameter• Stainless steel• Shaft length various• Shaft diameter various

Frequency Range

up to 15 kHz

up to 15 kHz

up to 15 kHz

Nominal Supply Voltage

9-18 VDC

10-25 VDC

8-30 VDC

Output Signal

One channel

Square Wave

Square Wave

Operating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 125°C

Typical Applications

Industrial, non demanding,
low cost applications

Industrial, non demanding,
low cost applications

Industrial, non demanding,
low cost applications

PAGE 55
SPEED SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

SPEED SENSORS
HALL EFFECT SPEED SENSORS

Jaquet DSY

Jaquet DSD 17

Jaquet DSD 40

Technology

Hall Effect chopped

Differential Hall Effect single
channel, 3 wires, voltage output

Differential Hall Effect single
channel, 2 wires, current output

Package• Stainless steel• Shaft length various• Various shaft diameter• Stainless steel• Shaft length various• Various shaft diameter• Stainless steel• Shaft length various• Various shaft diameter

Frequency Range

0 - 15 kHz

Up to 20 kHz

0 - 20 kHz

Nominal Supply Voltage

4.5 - 16 VDC
8 - 32 VDC

Nominal 15 VDC (9 VDC to 30 VDC)

Nominal 15 VDC (12 VDC to 30 VDC)

Output Signal

Square Wave

1 channel push-pull, voltage output

1 channel push-pull, current output

Operating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 125°C

Typical Applications

Industrial (Power Generation, Hydraulic,Engines, Industry)

Railway

Railway

VARIABLE RELUCTANCE SPEED SENSORS

Jaquet Green Line EV

Jaquet Green Line EX

Jaquet Green Line E

Jaquet SIL-3

Technology

Variable Reluctance (VR)square wave output

Variable Reluctance(VR) Classified Areas(explosion proof)

Variable Reluctance (VR)

Variable Reluctance (VR)

Package• Stainless steel• Shaft lenght various• Various shaft diameter• Stainless steel• Shaft lenght 48, 89, 129 mm• Shaft diameter 5/8"
and 3/4"• Stainless steel• Shaft lenght various• Shaft diameter various• Stainless steel• Shaft lenghts 35 mm
up to 101 mm• Shaft diameter M16 and 5/8"

Frequency Range

25 Hz - 20 kHz

25 Hz - 20 kHz

25 Hz - 20 kHz

up to 30 kHz

Nominal Supply Voltage

5-32 VDC

Passive

Passive

Passive

Output Signal

Square Wave

Sine Wave

Sine Wave

Sine Wave

Operating Temp.

-40°C to 125°C-40°C to 125°C-40°C to 125°C-40°C to 150°C

Typical Applications

Industrial, non demanding,
low cost applications

Industrial, non demanding,
low cost applications

Industrial, non demanding,
low cost applications

SIL-3 and SIL-4 applications

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 56
SPEED SENSORSSPEED SENSORS
POLE WHEEL

FTP520 One piece pole
wheel without boss

FTP530 One piece pole
wheel with boss

FTP540 & FTP560 Twopiece pole wheels

Material

1.1191 CK45 Ferromagnetic steel,electrogalvanized, whit/
blue passivated 8…12µm

1.1191 CK45 Ferromagnetic steel,electrogalvanized, whit/
blue passivated 8…12µm

1.1191 CK45 Ferromagnetic steel,electrogalvanized, whit/
blue passivated 8…12µm

Module1 to 4

1 to 3

1 to 3

Typical Applications

Raliway traction motors, turbines, diesel
engines, motors/generators and large
compressors in industrial machinery

Measuring chain/signal output optimized

Existing or new designed machine withdifficult mounting process of the pole wheel

FTP551 Pole BandFTP552 Pole BandFTP553 Pole BandFTP554 Pole Band

Material

Ferromagnetic Steel St 12.03,
surface Zinc-plated DIN/EN/
ISO 9227, passivated blue/
white 8-12µm

Ferromagnetic Steel St 12.03,
surface Zinc-plated DIN/EN/
ISO 9227, passivated blue/
white 8-12µm

Ferromagnetic Steel St 12.03,
surface Zinc-plated DIN/EN/
ISO 9227, passivated blue/
white 8-12µm

Ferromagnetic Steel St 12.03,
surface Zinc-plated DIN/EN/
ISO 9227, passivated blue/
white 8-12µm

Module

≥ 3.0

≥ 3.0

≥ 3.0

≥ 3.0

Typical Applications

Typically used for shafts withsmall diameters (diameter<500mm) and sensors
which are sensitive to highmagnetic gradients

Typically used for shafts withlarge diameters (diameter>500mm) and sensors
which are sensitive to highmagnetic gradients

Typically used for shafts withlarge diameters (diameter>500mm), large axial
movements of the shaft andlarge number of poles

Typically used for shafts withlarge diameters (diameter>500mm), large axial andradial movements of the shaft

PAGE 57
SPEED SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

SPEED SENSORSTACHOMETERS

T400 Tachometer

T500 dual channel Tachometer

T600 Multitasker

Analog Inputs

0

0

1

Binary Inputs

1

2

2

Analog Ouptuts

1

2

2

Relays1

44Communication Interface

RS232

LAN (TCP/IP)

LAN/CAN

Nominal Supply Voltage

10 to 36VDC

AC version: 90 to 264VAC
DC Version: 18 to 36VDC

AC version: 90 to 264VAC/120 to 370VDC
DC Version: 18 to 36VDC

Ambient Temperature-40°C to 85°C

-25°C to 50°C for AC version-40°C to 70°C for DC version-25°C to 50°C for AC version-40°C to 70°C for DC version

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 58
SPEED SENSORSTEMPERATURE SENSORSTEMPERATURESENSORS

PAGE 59
TEMPERATURE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

TEMPERATURE SENSORSSENSING ELEMENTS—NTC
Analog Output

Thermistor Chips

Radial Leaded
Thermistors

Axial Leaded Thermistors

Space Qualified (Hi-Rel)

Package

Leadless chips, SMD
0402, 0603, 0805

Radial, beads

DO-35

Radial, bead, custom

TypeGold or silver electrodes,
surface mounted

Epoxy or glass coated

Glass coated

NTC, epoxy, glass, probes

Resistance Range

Chip: 100 to 1MΩ/
SMD:2K to 200KΩ

100 to 1MΩ

5KΩ to 100KΩ

1KΩ to 100KΩ

Unique Features• Wire bonding compatible• End band SMD• Interchangeable• Moisture resistant• Stability• Tight tolerance (±1%)• Max. stability using highdensity (HD) chip
• Hermetically sealed• Tinned and nickel plated leads• ESA and NASA approved• High reliability and accuracy

Accuracy±1% to 10%

0.25% to 20%±1% to ±3%

0.5% to 10%

Operating Temp.

-40°C to 125°C-55°C to 280°C-40°C to 300°C-55°C to 160°C

Dimensions (mm)

Chip: 0.34 - 1 square
SMD 0402: 1 x 0.5 x 0.7
SMD 0603: 1.6 x 0.8 x 1
SMD 0805: 2 x 1.25 x 1.2

0.4 to 4.9

2.0 x 4.0 body

From 2.4

Typical Applications

Temperature compensation,communication (DWDM), infraredsensing systems, PCB mounting
temperature measurement

Temperature sensing for OEM,
automotive, medical, HVACR

Refrigeration including cabinet
sensing and evaporator coil,white goods, fire detection
units, air-conditioning systems,
PCB temp. sensing

Instrumentation and compensation
for aerospace applications

SENSING ELEMENTS—DIGITAL
Digital Output

Temperature System Sensor (TSYS)
Package

QFN16, TDFN8Type

I2C, SPI, PWM, SDM (Convertible to analog voltage)

Unique Features• Low power• Small size
• Calibrated and ready to use• 16-bit resolution

Accuracy

Up to ±0.1°C at -5°C to 50°C

Operating Temp.

-40°C to 125°C

Dimensions (mm)

QFN16: 4 x 4 x 0.85
TDFN8: 2.5 x 2.5 x 0.75

Typical Applications

Industrial control, replacement of precision RTDs, thermistors
and NTCs, heating and cooling systems, HVACR

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 60
TEMPERATURE SENSORSTEMPERATURE SENSORSSENSING ELEMENTS—RTD
Analog Output

Nickel RTD

Platinum Thin Film Chips

Package

SOT 23
Bare die on request

Leadless chips, SMD 1206Type

• Thin film nickel structure on silicon substrate, protected with apassivation layer• SOT 23 package for SMT• Bare die for COB assembly• Thin film platinum deposited on ceramic substrate• Contact pads on top and bottom side for NTC chip like assembly• Contact pads on both ends for SMT

Resistance Range

1,000Ω

100Ω, 1,000Ω (Other values on request)

Unique Features• Harsh environment compatible• Automotive qualified• Very small dimensions• Very short response time
• Good linearity• High temperature coefficient• Low power consumption• Good thermal connection of sensing
element through leadframe-pin• Long term stability• Interchangeability• Assembly like NTC chips• Very small dimensions• Short response time

Accuracy

Class B, according to former DIN 43760 standard

According to DIN EN 60751

Operating Temp.

-55°C to 160°C-50°C to 400 °C

Dimensions (mm)2.1 x 2.5 x 2.1 (SOT 23), 0.7 x 0.7 x 0.4 (Bare die)

1.5 x 1.5 (Top/bottom pads), 1.2 x 3.6 (SMT)

Typical Applications

Automotive, industrial, OEM, thermal compensation, thermal
management

White goods, automotive, industrial, aerospace,
medical, test and measurement

Platinum Thin Film Sensors

Glass Wire Wound Sensors

Ceramic Wire Wound Sensors

Package

Wired component

GO, GXCWW600, CWW850, CWW1000

Type• Thin film platinum deposited on
ceramic substrate, glass coated• Tube outline available• Connection via radial leads

Glass rod, radial leads

Ceramic rod, radial leads

Resistance Range

100Ω, 1,000Ω (Other values on request)

100Ω (2X 100Ω on few versions)100Ω (2X 100Ω on few versions)

Unique Features• Long term stability• Interchangeability• Small dimensions• Short response time
• High electrical insulation• Aggressive environments (Acid, oil, solvent)• Small dimensions• Stability• No hysteresis• Short response time
• Interchangeability• High temperature• Stability• No hysteresis• Small dimension• Interchangeability

Accuracy

Class T (F0.1), A (F0.15), B (F0.3)
according to DIN EN 60751

Class W0.3, W0.15, W0.1
according to IEC60751

Class W0.3, W0.15, W0.1
according to IEC60751

Operating Temp.

-50°C to 600°C (Standard) down to-200°C or up to 1,000°C (On request)-200°C to 400°C-200°C to 600°C (CWW600)-200°C to 850°C (CWW850)-200°C to 1000°C (CW1000)

Dimensions (mm)2.0 x 2.3 x 1.1 (Standard) 1.2 x 4.0 x 1.1(Standard) Other dimensions (On request)

Ø1.8/length 5 mm to Ø4.5/length 48 mm

Ø1.5/length 8 mm toØ4.5/length 30 mm
Ø2.7/length 45 mm (CWW1000)

Typical Applications

White goods, automotive, industrial,aerospace, medical, test and measurement

Oil and chemical industry, aviation,
aeronautic, food industry

Process industry, laboratories,
reference sensors

PAGE 61
TEMPERATURE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

TEMPERATURE SENSORSSENSOR ASSEMBLIES

Ring Sensors

Push-in Sensors

Screw-in Sensors

Refrigeration
Molded Probes

PackageRing for surface assemblyThreaded bolt, tube style

Brass, copper or stainlesssteel closed-end tube

Brass, copper or stainlesssteel housing withintegrated connector

PVC or TPE

Type

Epoxy potted element

Epoxy potted element, miniature design

Epoxy potted element,
rigid sheath

Overmolded

Sensor Range• NTC• RTD: Pt, Ni• NTC• RTD: Pt, Ni• Thermocouple: Type J, K, T, E

• NTC
• RTD: Pt, Ni, Cu• Thermocouple: Type J, K, T, E

• NTC
• RTD: Pt

Unique Features• Surface mount sensing• For use where space is limited• Simple installation• Corrosion resistant• Available with mounting tabs or clips• Corrosion resistant• Different thread types• Connectors available• Mounting clips available

Accuracy• NTC: Custom tolerances
available• Pt RTD: Class AA, A, B
according to IEC60751• NTC: Custom tolerances available• Pt RTD: Class AA, A, B
according to IEC60751• NTC: Custom tolerances
available• Pt RTD: Class AA, A, B
according to IEC60751• NTC: Custom tolerances
available• Pt RTD: Class AA, A, B
according to IEC60751

Operating Temp.

Varies: -50°C to 250°C

Varies: -50°C to 250°C

Varies: -50°C to 250°C-40°C to 125°C

Dimensions (mm)Case specific dimensions

Case specific dimensions

Custom lengths, diameters
and threads available

8 x 30, 6.5 x 25, 6 x 50,
6 x 5 x 15

Typical Applications

Surface plates, heatexchangers, fluid pumping
systems, generators

Boiler, liquid, evaporator, HVACR,
industrial processes control, district
heating and cooling, automotive,
bearing monitoring, motors, gear boxes

Boiler, liquid, HVACR, industrial
processes control, district
heating and cooling, immersion

HVACR, industrial
processes control

Pipe Mount Sensors

Outdoor Air Sensors

Pool and Spa Sensors

Boiler Sensors

Package

Copper or stainless steel housing

Metal housing with PVC sun shieldwith or without weatherproof box

Plastic or metal housingwith o-ring seal designed forband clamp or backing nut

Brass or SS housing

Type• Overmolded• Epoxy potted• Fast response time

• Overmolded subassembly• Threaded housing

Unique Features• Fast response time
• Moisture resistant construction• Easy installation – threads into
mounting hole or standard handy box• Fully potted housing protects
sensing element and provides
fast, accurate response• O-ring seals• Compatible with pooland spa chemicals• Integrated connector• Corrosion resistant• Different threads types
and connectors available

Accuracy• NTC: custom tolerances
available±0.2°C at 0°C to 70°C

±0.2°C• NTC: Custom tolerances
available• Pt RTD: Class AA, A, B
according to IEC60751

Operating Temp.

-40°C to 125°C-40°C to 105°C

0°C to 90°C

Varies: -50°C to 250°C

Dimensions (mm)Custom configurations available

Ø12 X 64

6.4 x 50

Custom lengths, diameters
and threads available

Typical Applications

Industrial process, boiler control,
HVACR, refrigeration, foodservice, energy management,
test equipment

Residential and commercial
building controls, energy
management systems

Pools, hot tubs

Boiler control, liquid, industrial
processes control, district
heating and cooling, immersion

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 62
TEMPERATURE SENSORSTEMPERATURE SENSORSSENSOR ASSEMBLIES

Oven Sensors

Urea Temperature Sensors

Exhaust Gas Temperature Probes

Package

Stainless steel housing

Plastic housing with screw hole mountings

EGT thermocouple probeType• Pt element encapsulated into ceramic
tube, with rigid stainless steel housing• High temperature cable• Overmolded plastic housing withintegrated 2 pin connector• Mineral insulated alloy sheath, screwed
mechanical interface, cable extension
and automotive connector• Option: CANbus interface (From 1 to 4
thermocouples, fully configurable)

Sensor Range

Pt100, Pt500, Pt1000 sensor

NTC

Thermocouple: Type K, N

Unique Features• High temperature• Easy integration/installation• Higher dielectric strength according to type

• Temperature measurement of urea liquid
used in Selective Catalytic Reduction(SCR) systems• Suitable for high pressure applications• High temperature, robust design• Vibration and corrosion resistant• Fast response time

Accuracy

Class B, C according to IEC60751• NTC: custom tolerances available• ±2%, 3% and 5%• Beta 25/85: 3976

Class 1 according to IEC584

Operating Temp.

-20°C to 750°C (According to version)-40°C to 125°C-40°C to 900°C

Dimensions (mm)• OD Ø4 mm to Ø6 mm• Immersion length 35 mm to 100 mm• Custom mechanical interface andcable length

Sensor tip 8 mm diameter• OD Ø4 to OD Ø8• Custom immersion length and cable length

Typical Applications

Drying oven, domestic oven

Temperature measurement of urea
liquid used in SCR systems

Automotive, truck, mining, power unit, racing

Micro-Thermocouples

Patient Monitoring Probes

TLH Reference Probe

Package

Fine gage thermocouples

Sensor with cable and connector

TLH100/TLH600

Push-in probe with handle

Type• Micro sized thermocouple:
44 AWG, 40 AWG,
38 AWG, 36 AWG• Polymer encapsulated
or bare junction• Reusable: Skin; 10FR and
12FR GP
Disposable: Skin; 9FR and
12FR GP;
12FR, 18FR, 24FR Esoph/
Stethoscope;
14FR, 16FR, 18FR Foley catheter• Rigid protective external
stainless steel sheath andstainless steel handle, uniqueinternal design to insure stability• Versatile push-in probe withstainless steel sheath and plastic
or stainless steel handle• High precision sensing element
combined with integrated
electronics for signal processing,calibration and USB interface

Sensor Range

Thermocouple type: T, K

400 series, 700 series(Reusable only)

Pt100 sensor

Not applicable due to
direct digital output

Unique Features• Welded or soldered junction• Low profile, fast response• Polyesterimide wire insulation• Autoclavable reusables• Sterile disposables• Stability• Provided with calibration
report or option of calibration
certificate by national
committee for accreditation(COFRAC)• USB conformal interface• Calibrated digital output,
recalibration possible on request• Robust design for general
purpose applications• Long term stability

Accuracy

Varies by type: standard,
special and custom limitsof error available±0.1°C at 25°C to 45°C
ISO-80601-2-56: ±0.2°C at
35°C to 42°C

Class B (TLH600), A (LTH100)
according to IEC60751±0.1°C for temperature range-5°C to 55°C
±0.2°C for temperature range-40°C to 160°C(Other accuracies on request)

Operating Temp.

Varies by type: Rated
up to 240°C-40°C to 100°C,
Patient: 0°C to 50°C

-80°C to 350°C (TLH100)-180°C to 600°C (TLH600)-55°C to 160°C for probe tip-40°C to 85°C for handle withelectronics (Other temperature
ranges on request)

Dimensions (mm)Varies by thermocouple gage

Reusable: 3 m cable with sensor
Disposable: Sensor <1 m; 3
m reusable adaptor cableOD Ø5 x 500 + handle Ø15 x 100(Typical cable length = 2 m)

OD Ø6 x 200 + handle Ø19 x 100(Typical cable length = 2,000)

Typical Applications

Medical, catheters

Patient monitoring, laboratory

Laboratory, temperature sensors
calibration by comparison

Laboratory, mobile research,
test and measurement

PAGE 63
TEMPERATURE SENSORS

te.com/sensors

USB Temperature Probe

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

TEMPERATURE SENSORSSENSOR ASSEMBLIESStator Sensors

Surface Sensors

Bearing Sensors

Package

TPE/CPME
G11 epoxy glass laminated, Class F or H ilicone rubber or polyimideS
laminated element
SP683

Copper alloy tip
Stainless steel, isolated stainlesssteel or epoxy glass caseType

• Rigid flat, slot sensor• Cable or leadwire options• Flat, flexible, rectangular sensor• Variety of designs available• Rigid sheath• Tip sensitive• Cable/leadwire options

Sensor Range• RTD: Pt, Ni, Cu• Thermocouple: Type J, K, T, E

• RTD: Pt, Ni, Cu• Thermocouple: Type J, K, T, E

• RTD: Pt, Ni, Cu• Thermocouple: Type J, K, T, E

Unique Features• Extended sensitive length• Single or dual elements• Calibration available• Surface sensing for curvedor uneven surfaces• Noninvasive, simple installation• Adhesive backing option• Cut-to-length• Copper tip for fast time response• Assemblies with fluid sealand spring loading• Single or dual elements

Accuracy

RTD: Class A, B according to IEC60751

RTD: Class A, B according to IEC60751

RTD: Class A, B, C according to IEC60751

Operating Temp.

Max. temperature: Class F, 155°C
Max. temperature: Class H, 180°C
Available up to 200°C

Varies: -50°C to 200°C Available up to 220°C

Sheath specific, up to 250°C

Dimensions (mm)Custom dimensions available

Custom dimensions available

Custom lengths Standard sheath
diameters: Ø4.78, Ø5.46, Ø6.35

Typical Applications

Monitor temperature between stator
coils, electric motors, generators

Chemical and pharmaceutical industry,
process industry, laboratory, aerospace, motor
end windings of stator coils, generators

Bearing monitoring, electric motors,
generators

Thermocouple

Transmitter

PackageScrew-in or push-in design with cable extension, connector, orconnecting head

Brass, copper and stainless steel housing, flexible sheath withintegrated connector.

Type• Collapsible Mineral Insulated (MI) with alloy sheath (Radius ≥5*OD)• Flexible cable with plastic or composite insulation• Rigid protection sheath: ceramic, quartz or alloy sheath• Epoxy potted element• Screw-in

Sensor Range

Type T, J, K, N, R, S, B (According to TC type and insulation type)

4 - 20 mA output

Unique Features• High temperature and high vibration level (For MI)• Available in small diameters for fast respond time
• Grounded or ungrounded or apparent hot junction• Single or multiple measuring points• Compact, welded design• Highly sensitive and stable• High vibration application• Good waterproof properties

Accuracy

Class 1 according to IEC584

0.5 or 1% FS

Operating Temp.

-40°C to 1,700°C (According to TC type and insulation type)-20°C to 120°C

Dimensions (mm)• OD Ø0.3 mm to Ø8 mm for MI• Ø0.15 mm for smallest flexible cable• Custom dimensions, fittings and cable lengths(From few centimeters to many meters)• Customer sheath length, thread type
• Probe diameter: Ø4.75 mm; Ø5 mm; Ø6 mm; Ø6.35 mm; Ø8 mm

Typical Applications

Aeronautic, process industry, medical, semiconductor industry(spike, profile)

Heavy industry, general industrial monitoring

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 64
TEMPERATURE SENSORSTEMPERATURE SENSORSTHERMOPILES

TS

TSD

Package

TO-18, TO-5

TO-5

Type

Thermopile sensor components

Digital thermopile sensor component

Temp. Range

Depends on applied electronics and calibration, filter
types optimal for object temperature range -40°C
to 300°C (Extended range: -60°C to 1,000°C)

Object temperature range 0°C to 300°C (Other
temperature ranges available upon request)

Unique Features• High signal output• Accurate reference sensors• Calibrated and ready to use, I2C interface• Direct assembly to PCB, no additional components needed

Accuracy

Depends on applied electronics and calibration

Depends on temperature range, typical 1% full range

Operating Temp.

Ambient temperature range: -20°C to 85°C

Ambient temperature range: -20°C to +85°C

Dimensions (mm)Ø9.15 x 4.4 (Body)

Ø9.15 x 4.4 (Body)

Typical Applications

Medical thermometer (ear, forehead), pyrometerContactless temperature measurement, e.g. on moving parts likeheated rolls, laminators, people detection, body temperature,
microwave oven, air conditioner

TS318-3B0814, TS318-5C50, TS305-10C50

PAGE 65
TEMPERATURE SENSORS

te.com/sensors

Single Pixel Digital Output SeriesSpecifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

TEMPERATURE SENSORSTHERMOPILES

TSEV

TSEV

TPT

Package

OEM-module

OEM-module

IP65 stainless steel tube

Type

Single-pixel thermopile module

8-pixel-linear array thermopile module

Thermopile system for industrial use

Temp. Range

Object temperature range 0°C to 300°C(Other temperature ranges available
upon request)

Object temperature range -20°C to 120°C

Object temperature range 0°C to 300°C

Unique Features• Calibrated, Interfaces: I2C, SPI• Different field of views:• 5° at 50%, 10° at 50%, 90° at
50%, others on request• Calibrated and ready to use• Digital output• Small field of view• Calibrated and ready to use• Digital or analog outputs• Small field of view

Accuracy

Depends on temperature range, typical
1% full scale, max. accuracy 0.1°C

Depends on temperature range, typical
2% full scale

Depends on temperature range,
typical 1% full scale

Operating Temp.

Ambient temperature range: 0°C to 85°C

Ambient temperature range: -20°C to 85°C

Ambient temperature range: 0°C to 85°C

Dimensions (mm)35 x 25 x 13 to 31

25 x 35 x 15.2

Ø18 x 111

Typical Applications

Contactless temperature measurement, e.g.
on moving parts or heated rolls, laminators,
people detection, microwave oven, airconditioner

Contactless temperature measurement, e.g.
on moving parts or heated rolls, laminators,
people detection, microwave oven, airconditioner

Contactless temperature measurement,
e.g. on moving parts or heated rolls,
control of assembly lines, paper
fabrication, drying applications

Single Pixel Series

SS-TS-TE101
06/2019

Multi Pixel SeriesSpecifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

TPT300V

PAGE 66
TEMPERATURE SENSORSTORQUE SENSORSTORQUESENSORSSTATIC TORQUE SENSORSCS1060CS1120CS1210

Package

Square male coupling

Keyed shaft connections

Collar mechanical fittings

Operating Mode

Reaction

Reaction

Reaction

Unique Features• Optional high level output• Keyed shaft connection• Static measurements• Optional high level output• Static torque measurement• Excellent temperature stability• High stiffness• Low transverse sensitivity• Optional high temp option

FS Ranges±5 to ±7,000 Nm±4 to ±5,600 lbf-ft±5 to ±2,500 Nm±4 to ±2,000 lbf-ft±160 to ±10,000 Nm±128 to ±8,000 lbf-ft

Max. Over-range

1.5X FS

1.5X FS

1.5X FS

Output/Span±20 mV , 0.5-4.5V±20 mV , 0.5-4.5V±20 mV , 0.5-4.5V

Combined Nonlinearity & Hysteresis< ±0.25% FS< ±0.25% FS< ±0.25% FS

Operating Temp.

-20°C to 100°C-20°C to 100°C-40°C to 150°C

Dimensions (mm)Application dependent

Application dependent

Application dependent

Typical Applications

Non-rotating parts torque measurement,
robotics and effectors,
laboratory and researchNon-rotating parts torque measurement,
robotics and effectors,
laboratory and researchNon-rotating parts torque measurement,
robotics and effectors,
laboratory and research

PAGE 67
TORQUE SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

TORQUE SENSORSDYNAMIC TORQUE SENSORS

CD1050CD1095

CD1110

Package

Square male couplings

Keyed shaft couplings

Keyed shaft couplings

Operating Mode

Dynamic rotary

Dynamic rotary

Dynamic rotary

Unique Features• Optional high level output• Rugged• Slip ring measurement interface• High accuracy• Built-in amplifier• Bi-directional measurement• Low range measurements• Bi-directional measurement• Mechanical over-range stops

FS Ranges±5 to ±7,000 Nm±4 to ±5,600 lbf-ft±5 to ±2,500 Nm±4 to ±2,000 lbf-ft±0.05 to ±2 Nm±0.04 to ±1.6 lbf-ft

Max. Over-range

1.5X FS

1.5X FS

10X FS

Output/Span±20 mV, 0.5-4.5V±20 mV, 0.5-4.5V±20 mV, 0.5-4.5V

Combined Nonlinearity & Hysteresis< ±0.25% FS<±0.25% FS<±0.25% FS

Operating Temp.

-20°C to 80°C

-20°C to 80°C

-20°C to 80°C

Dimensions (mm)Application dependent

Application dependent

Application dependent

Typical Applications

Engine efficiency, robotics andeffectors, laboratory and research

Process control equipment, robotics
and effectors, test and measurement

Process control equipment, robotics
and effectors, test and measurement

CONTACTLESS TORQUE SENSORS

CD1140CD9515

Package

Keyed shaft couplings

Keyed shaft couplings

Operating Mode

Contactless

Contactless

Unique Features• High accuracy• Built-in amplifier• Speed and angle
detection• Economical• Small form factor• Speed and angle
detection

FS Ranges±0.05 to ±20,000 Nm±0.04 to ±16,000 lbf-ft±5 to ±1,000 Nm±4 to ±738 lbf-ft

Max. Over-range

2X FS

2X FS

Output/Span±10 V (60 pulses/rev)±5 V (60 pulses/rev)

Combined
Non-linearity& Hysteresis±0.1% FS

±0.3% FS

Operating Temp.

0°C to 60°C

0°C to 60°C

Dimensions (mm)Application dependent

Application dependent

TypicalApplications

Process control
equipment, robotics
and effectors, test
and measurement

Industrial applications,process control
equipment, PLC
compatible

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

AUTOMOTIVE TEST TORQUE SENSORSFCA7300

CD1124T

Package

Steering wheel adaptable

Engine shaft dynamic
torquemeter

Operating Mode

Multi-sensing

Contactless

Unique Features• Dual torque/angle range• Steering velocity
measurement• Fits all road vehicles• Heavy duty vehicles• Telemetry based• Gearbox to engine
shaft measurement

FS Ranges

10 to 200 Nm(7 lbf-ft to 150 lbf-ft)

20,000 Nm
16,000 lbf-ft

Max. Over-range

10X FS

1.5X FS

Output/Span

±10 V

±10 V

Combined
Non-linearity& Hysteresis±0.1% FS

±1% FS

Operating Temp.

-20°C to 80°C

-20°C to 80°C

Dimensions (mm)Ø195 x 50

Ø195 x 35

TypicalApplications

On-car road test, truck andbuses steering test, armoredvehicles steering test

Automotive test
benches for engine shaft
torque measurementte.com/sensors

PAGE 68
TORQUE SENSORSULTRASONIC SENSORSULTRASONICSENSORS(air bubble, point level, continuous level monitoring)

STANDARD CONTACT POINT LEVEL

LL-01

LL-10

Type

Gap

Tip

Unique Features• All 316L SS• Integral electronics• Miniature threads• No adjustment for viscosity, density• All 316L SS• Integral electronics• No adjustment for viscosity, density

Input

5 - 30 VDC

5 - 30 VDC

Output• 30 V, 3 W relay• Analog 4 - 20 mA power loop

• 1 A SPDT
• Analog 4 - 20 mA power loop

Pressure Range

250 psi1000 psi

Operating Temp.

-30°C to 80°C

-30°C to 80°C

Actuation point

0.25 inches

Custom (2.25, 6, 12, 18, 24 inches)

Process Connection

1/4"NPT and 1/2"NPT

3/4"NPT

Cable

1, 4, 10, 20 feet

1, 4, 10, 20 feet

Approvals

CE

CE

Typical Applications

Medical waste tanks, histology processors,compressors, chillers, coolant reservoirs

Hydraulic reservoirs, storage tanks, pipe lines, sewage systems

PAGE 69ULTRASONIC SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

ULTRASONIC SENSORS
AIR-BUBBLE AND NON-INVASIVE POINT LEVEL

AD-101

SL-630Type

Non-invasive

Non-invasive

Unique Features• Detect minimum bubble size of 70% ID• Immune to EMI/RFI• Acoustic coupling agent not required• Continuous self testing• LED indicator• Stick on dry contact• Point level detection

Input

5 - 24 VDC standard

5 - 24 VDC

OutputTTL & Open collector

TTL (High), dry condition

Pressure Range

Atmosphere

Atmosphere

Operating Temp.

0°C to 40°C

-30°C to 70°C

Actuation point—

Variable

Process Connection—

Reusable sensor, disposable tape

Cable (Inches)

12

12

Approvals-

CE

Typical Applications

Infusion pumps, dialysis machines, semi-conductor equipment,
3D printing

Chromatography, chemical analyzer, hemodialysis, reagent vessels

CONTACT MULTI-POINT LEVELCONTINUOUS LEVEL

SL-900ML

TypeContact

Type

Continuous transmitter through air

Unique Features• Miniature• 10 µRA electropolished finish• 316 LSS body
• Designed for high purity market

Unique Features• Non-contact• Remotely mounted• 316 SS or epoxy sensor material• Configurable via RS-232

Input

VariableInput

24 VDC

Output

RS-232, analog, relay setpoints

Pressure Range

Atmosphere

Operating Temp.

-30°C to 70°C

Sensing Range

0.5" to 5" inches

Process Connection—

Accuracy±0.0075"

Elect Connection

Terminal block

OutputDual color LED and ½ A relay

Pressure Range

250 PSIG

Operating Temp.

-30°C to 93°C

Actuation point

Variable

Process Connection

1/2", 3/4" VCR, male/female

Cable (Inches)

Up to 24” shielded with strain
relief, 9 pin connector

Approvals

NEMA 1 housing

Approvals

NEMA 1 housing

Typical Applications

Pharmaceutical and semiconductor
industries, high pressure vessels

Typical Applications

Microplate well level, test tubes and vials,
bottle fill level, surface flaw detection

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 70ULTRASONIC SENSORSVIBRATION SENSORSVIBRATIONSENSORS

PAGE 71
VIBRATION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

VIBRATION SENSORSEMBEDDED ACCELEROMETERSMEMS DC Accelerometer

3022, 3028

3052A, 3058A

3038

3255A

PackagePins or pads

Pins or pads

SMD

SMD

Type

Board level

Board level

Board level

Board level

FS Range±2g to ±100g±2g to ±100g±50g to ±6,000g±25g to ±100g

Unique Features• mV output, critically gas damped• Board and screw mount options• Pin or solder pad option• Temperature compensated• Board and screw mount options• Pin or solder pad option• Hermetically sealed• High over-range protection• Gas damping• Amplified, signal conditioned• Gas damping• Bidirectional mounting

Accuracy±1.0% non-linearity±1.0% non-linearity±1.0% non-linearity±1.0% non-linearity

Operating Temp.

-40°C to 125°C-40°C to 125°C-54°C to 125°C-40°C to 125°C

Dimensions (mm)22.8 x 15.2 x 5.3

22.8 x 15.2 x 5.3

7.5 x 7.5 x 3.3

13.5 x 7.6 x 3.8

Typical Applications

Vibration and shock monitoring,tilt applications, motion
control, impact testing

Vibration and shock monitoring,tilt applications, motion
control, impact testing

Vibration and shock monitoring,embedded systems, shock
testing, safe and arm

Vibration and shock monitoring,aerospace testing, impact
testing, transportation

EMBEDDED ACCELEROMETERS
Piezoelectric Accelerometer

805, 805M1

808, 808M1

820M1

832, 832M1

834, 834M1

Package

TO-5

TO-8

Board level

SMD

SMD

Type

Adhesive (Stud
mount option)

Adhesive (Stud
mount option)

SMD

Board mount

Board mountFS Range±20g to ±500g±4g to ±50g

±25g to ±500g±25g to ±500g±2,000g to ±6,000g

Unique Features• Hermetically sealed• Case grounded design• Bandwidth to 12 kHz• Hermetically sealed• Case grounded design• Bandwidth to 8 kHz• Small size, low cost
• Dynamic response• Wide >6000Hz
bandwidth• SMT mount, triaxial• Wide bandwidth >6000Hz• Low power consumption• SMT mount, triaxial• Wide bandwidth >6,000Hz• Low power consumption

Accuracy±1.0% non-linearity±1.0% non-linearity±2.0% non-linearity±2.0% non-linearity±2.0% non-linearity

Operating Temp.

-50°C to 125°C-50°C to 100°C-40°C to 125°C-20°C to 80°C (832)-40°C to 125°C (832M1)-20°C to 80°C (834)-40°C to 125°C (834M1)

Dimensions (mm)Ø8.9 x 10.2

Ø15.2 x 16.6

8.9 x 8.9 x 4.2

18.8 x 14.2 x 4.3

18.8 x 14.2 x 4.3

Typical Applications

Machine monitoring,data loggers,
permanent structures

Machine monitoring,data loggers, embedded
applications

Embedded predictive
maintenance, condition
monitoring, data loggers

Data logging, asset
monitoring, impact
monitoring

Data logging, asset
monitoring, impact
monitoring

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 72
VIBRATION SENSORSVIBRATION SENSORSPLUG AND PLAY ACCELEROMETERS
Uniaxial DC Response

40A, 40B

52, 52M30, 52F

58

Package

Anodized aluminum

Anodized aluminum

Anodized Aluminum

Type

Screw mount

Adhesive and Screw mount

Adhesive mountFS Range±25g to ±2,000g±50g to ±6,000g±50g to ±6,000g

Unique Features• Critically damped• Euro NCAP certified• Compact• SAE-J211/2570 and ISO-6487 compliant• Gas damping, thin profile• Over-range stops• Most reliable crush zone
accelerometer availbale• Rugged, water proof• Mounting on any sides of housing

Accuracy±1.0% non-linearity±1.0% non-linearity±1.0% non-linearity

Operating Temp.

-20°C to 80°C

-40°C to 90°C

-20°C to 85°C

Dimensions (mm)16.7 x 10.0 x 5.0

11.2 x 10.1 x 3.8

14.0 x 6.3 x 6.3

Typical Applications

Pedestrian crash testing, Euro NCAP testing

Vibration and shock testing, safety
impact testing, side-impact testingOn-vehicle crash and impact testing,drop testing, harsh environment

64B, 64C, 64X

1201, 1201F

Package

Anodized aluminum

Anodized aluminum

Type

Screw mount

Adhesive/screw mountFS Range±50g to ±6,000g±50g to ±2,000g

Unique Features• SAE-J211/2570 and ISO-6487 compliant• ATD dummy certified• Market performance leader• Cube form factor, low noise cable• Adhesive or screw mount• Over-range stops

Accuracy±0.7% non-linearity±1.0% non-linearity

Operating Temp.

-40°C to 121°C-20°C to 85°C

Dimensions (mm)13.1 x 10.0 x 5.0

8.9 x 8.9 x 9.4

Typical Applications

In-dummy crash and impact testing, Euro NCAP testingOn-vehicle crash and impact testing, vibration and shock monitoring

PAGE 73
VIBRATION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

VIBRATION SENSORSPLUG AND PLAY ACCELEROMETERS
Uniaxial DC Response

3801A, 4801A

3700

EGAXT, EGAXT3

Package

Stainless steel

Stainless steel

Stainless steel

TypeStud mountScrew mount

Adhesive/screw mountFS Range

4801A; ±2g to ±200g
3801A; ±50g to ±6,000g±50g to ±6,000g±5g to ±2,500g

Unique Features• Hermetically sealed sensor• mV and amplified output options• Integral connector, detacheable cable• No zero shift• mV output• 20,000 g over-range protection• Sub-miniature, fluid damped• Miniature single and triaxial designs• 10,000 g over-range protection

Accuracy±0.1% non-linearity on 4801A±1.0% non-linearity±1.0% non-linearity

Operating Temp.

-54°C to 121°C-54°C to 121°C-40°C to 120°C

Dimensions (mm)15.9 x 15.2

14.2 x 8.1 x 3.8

EGAXT; 7.2 x 4.6 x 4.6
EGAXT3; 12.7 x 12.7 x 12.7

Typical Applications

Impact testing, structural testing, test andinstrumentation, environmental testing

Impact and shock testing, structural
testing, drop testing, aerospace testing

Flight test and control, launch,crash, impact testing, robotics

EGCS-D0, EGCS-D1S, EGCS-D5

4602, 4604

4610, 4810A

Package

Stainless steel

Anodized aluminum

Stainless steel

Type

Screw/stud mountScrew mountScrew mountFS Range±5g to ±10,000g±2g to ±200g±2g to ±200g

Unique Features• Rugged housing• Critically damped• 20,000 g over-range protection• UltraStable MEMS
• Low noise, signal conditioned• <2% TEB (total error band)• UltraStable MEMS
• Hermetically sealed• <2% TEB (total error band)

Accuracy±1.0% non-linearity±0.1% non-linearity±0.1% non-linearity

Operating Temp.

-40°C to 120°C-54°C to 125°C-55°C to 125°C

Dimensions (mm)

D0: 19.0 x 19.0 x 7.6
D1S: 12.7 x 12.7 x 15.2
D5; 14.2 x 12.7 x 5.6

21.1 x 21.6 x 7.6

25.4 x 29.1 x 7.6

Typical Applications

Impact and shock testing, destructive
testing, engine testing

Flight testing on engines, flutter test,road load and transportation testing

Low frequency monitoring, roadtesting, motion analysis

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 74
VIBRATION SENSORSVIBRATION SENSORSPLUG AND PLAY ACCELEROMETERS
Triaxial DC Response

53A, 53AF

68CM1

4630

4020, 4030

606M1

Package

Anodized aluminum

Stainless steel

Anodized aluminum

Molded plastic

Nitrile rubber pad

TypeAdhesive mountScrew mountScrew mountScrew mountRemovable

FS Range±50 to ±2,000g±50 to ±2,000g±2g to ±200g±2g to ±6g±25g

Unique Features• Low cost
• Gas damping• Low power• World SID• Gas damping• Low power• UltraStable MEMS
• Low noise, signal
conditioned• <2% TEB (total
error band)• Low cost
• Biaxial, withtriaxial option• DC response• 606M2 IEPE option• Triaxial, hermetic• Seat pad accelerometer

Accuracy±1.0% non-linearity±1.0% non-linearity±0.1% non-linearity±1.0% non-linearity±1.0% non-linearity

Operating Temp.

-20°C to 85°C

-20°C to 85°C

-40°C to 115°C-40°C to 85°C

-20°C to 85°C

Dimensions (mm)18.3 x 13.2 x 7.1

12.7 x 12.7 x 12.7

26.2 x 26.2 x 23.4

71.2 x 40.0 x 15.2

199 x 4

Unique Features

Auto safety, passenger
comfort, transportation,
NVH analysis

Auto safety, in-dummy
crash, on-vehicle crash

Road testing, motion
control, structural testing

Structural monitoring,seismic array,
bridge testing

Off road equipment,
amusement rides,
commercial aircraft

4835A

4630M12, 4630M14

4332M3

XL403D

PackageTitanium

Titanium

Stainless steel

Nickel plated aluminum

Type

Screw mountScrew mountScrew mountScrew mountFS Range±2g to ±200g±2g to ±200g±2g to ±50g

±2g to ±15g

Unique Features• UltraStable MEMS
• Welded connector, hermetic• <2% TEB (total error band)• UltraStable MEMS
• Compact, rugged anddouble shielded cable• <2% TEB (total error band)• Low noise ranges• Temperature compensated• High over-range• Digital triaxial accelerometer• Smart, onboard processing• Temperature output included

Accuracy±0.1% non-linearity±0.1% non-linearity±1.0% non-linearity±0.25% non-linearity

Operating Temp.

-55°C to 125°C-55°C to 125°C-40°C to 115°C-40°C to 85°C

Dimensions (mm)22.9 x 22.9 x 17.1

22.9 x 22.9 x 16.0

34.5 x 34.5 x 31.2

36.5 x 25.4 x 17.5

Typical Applications

Road testing, motion
control, flight testing

Road load testing,transportation testing

Structural monitoring,bridge testing

Smart sensor function, vibration
monitoring and alarm functions

PAGE 75
VIBRATION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

VIBRATION SENSORSPLUG AND PLAY ACCELEROMETERSIEPE AC Response

7100A, 7101A

7102A

7108A

Package

Stainless steel/titanium

Titanium

Stainless steel

TypeCenter-hole mount

Adhesive mount

Adhesive mounting

FS Range±50g to ±2,000g±50g to ±2,000g±50g to ±500g

Unique Features• Single axis, shear mode
• Isolated mounting surface• Wide bandwidth, >10 kHz• Single axis, shear mode
• Wide bandwidth• Single axis, shear mode
• Wide bandwidth• Small size

Operating Temp.

7100A: -55°C to 150°C
7101A: -55°C to 125°C-55°C to +125°C-55°C to 125°C

Dimensions (mm)7100A: 9.9 x 22.3, 7101A: 5.8 x 14.5

4.4 x 11.9

9.5 x 10.2

Typical Applications

Flight testing, general purpose,
vibration monitoring

Small structures monitoring, minimalmass loading, general purpose testing

Vibration monitoring, modal
testing, general purpose

7104A, 7105A

7131A, 7132A

7135A

Package

Stainless steel

Titanium

Stainless steel

TypeStud mounting

Adhesive/stud mounting

Adhesive mounting

FS Range±50g to ±2,000g±50g to ±2,000g±50g to ±500g

Unique Features• Single axis, shear mode
• Wide bandwidth• Triaxial, shear mode
• >12 kHz bandwidth• Hermetically sealed• Triaxial, through hole mount• Case isolated, internally shielded• Hermetically sealed

Operating Temp.

-55°C to 125°C-55°C to 125°C-55°C to 125°C

Dimensions (mm)7104A: 11.11 x 14.10 ,7105A: 11.11 x 19.05

7131A: 11.0 x 11.0 x 11.0, 7132A: 15.2 x 15.2 x 14.5

28.6 x 14.0

Typical Applications

General purpose IEPE accel, vibration
monitoring, lab testing

General purpose, modal testing,vibration monitoring

AD&M monitoring, HUMS, structural
applications

8711-01

8011-01, 8021-01

8032-01

8811-01

Package

Stainless steel

Stainless steel

Stainless steel

Stainless steel

TypeStud mount

Stud/center-hole mountStud mountStud mountFS Range±5g to ±500g±10g to ±100g±50g to ±500g±5g to ±500g

Unique Features• Industrial accelerometer• Case isolated, internal shielding• Low cost

• Industrial accelerometer• Case isolated, internal shielding• Reverse wiring protection• Industrial accelerometer• Case isolated, internal shielding• Low cost, molded strain relief• Certified for wind tubines• ±2,500VAC lightning protection• Case isolated, internal shielding

Operating Temp.

-55°C to +125°C-55°C to 125°C-40°C to 100°C-55°C to +125°C

Dimensions (mm)22.2 x 50.8

22.2 x 48.3

14.3 x 45.3

22.2 x 50.8

Typical Applications

Industrial applications, machinemonitoring, wind turbines

Industrial applications, machinemonitoring, intrinsic safety

Industrial applications,
machine monitoring

Industrial applications, machinemonitoring, wind turbines

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 76
VIBRATION SENSORSVIBRATION SENSORSPLUG AND PLAY ACCELEROMETERS
4-20mA AC Accelerometer

8011, 8021-AR/AP

8011, 8021-VR/VP

Package

Stainless steel

Stainless steel

TypeStud/center-hole mount

Stud/center-hole mountFS Range

5g to 50g

0.5in/sec to 5.0in/sec

Unique Features• Industrial 4-20mA accelerometer• Case isolated, internal shielding• Top mount or side mount connector• Industiral 4-20mA velocity transmitter• Case isolated, internal shielding• Top mount or side mount connector

Operating Temp.

-40°C to 85°C

-40°C to 85°C

Dimensions (mm)22.2 x 48.3

22.2 x 48.3

Typical Applications

Industrial applications, machine monitoring, intrinsic safety

Industrial applications, machine monitoring, intrinsic safety

PLUG AND PLAY ACCELEROMETERS
3 Wire Voltage AC Accelerometer

8102A
Package

Anodized aluminum

Type

Screw mountFS Range±25g to ±6,000g

Unique Features• Triaxial PE accel, low cost
• Environmentally sealed, rugged• <22µA current consumption

Operating Temp.

-40°C to 125°C

Dimensions (mm)25.4 x 21.6 x 10.8

Typical Applications

Impact and shock testing, R&D and lab applications

PAGE 77
VIBRATION SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

VIBRATION SENSORSPLUG AND PLAY ACCELEROMETERS
PE Charge Accelerometer

7500A

7501A

7502A

Package

Stainless steel

Titanium

Titanium

Type

Center-hole mountCenter-hole mount

Adhesive mounting

Sensitivity

7pC/g to 20pC/g

5.6pC/g

1.8pC/g

Unique Features• Single axis, shear mode
• Hermetically sealed• Isolated mounting surface• Single axis, shear mode
• Hermetically sealed• Bandwidth to >15 kHz• Single axis, shear mode
• Hermetically sealed• Miniature, <1 gram

Operating Temp.

-73°C to 260°C-73°C to 260°C-73°C to 260°C

Dimensions (mm)8.4 x 22.3

5.8 x 14.5

4.4 x 11.9

Typical Applications

Gearbox vibration monitoring, flight
test, high temp. applications

Gearbox vibration monitoring, flight
test, high temp. applications

Small structures monitoring, minimalmass loading, high temp. applications

7504A, 7505A

7514A

7531A

Package

Stainless steel

Stainless steel

Titanium

Type

Stud mountStud mounting

Adhesive mountSensitivity

5.6pC/g

30pC/g to 100pC/g

1.8pC/g

Unique Features• Single axis, shear mode
• Top and side connector option• >15 kHz Bandwidth• Single axis, shear mode
• >12 kHz bandwidth• High sensitivity• Triaxial, shear mode
• Miniature, light weight• >10 kHz bandwidth

Operating Temp.

-73°C to 260°C-73°C to 260°C-73°C to 260°C

Dimensions (mm)7504A; 11.1 x 14.1, 7505A; 11.1 x 19.0

15.0 x 15.0

11.0 x 13.6 x 11.0

Typical Applications

Small structures monitoring, general
purpose, high temp. applications

Low frequency vibration, general purpose,high temp. applications

High temp. applications, flight testing,structural monitoring

PLUG AND PLAY ACCELEROMETERS
Amplifiers

121

130

140A & 142A

160

161

Type

Bench top

In-line charge converter

Inline amplifier with auto-zeroBench top

Bench top

No. of Channels

3

1

1

1

4

Gain Range

x0.001 to 9999

x0.1, 1, 10

x10, 25, 50, 100, 200

x1, 10x0.001 to 999.9

Unique Features• Universal DC amplifier• Low noise operation with
auto-zero• For bridge type sensors• Low noise, small package• BNC male or female• Wide bandwidth• ±1.5 mV auto-zero• 140A for bridge type sensor,
142A for strain gages• 5 to 30 VDC excitation• Economical IEPE
power supply• Portable, compact• Rechargeable battery• Charge and IEPE
conditioner• Sensitivity normalization• Support IEEE 1451.4 TEDS

Dimensions (mm)301 x 258 x 102

Ø13.8 x 52.2

57 x 25 x 13

101 x 83 x 32

310 x 180 x 115

Typical Applications

Instrumentation labs, test
benches, R&D facilities

Instrumentation labs,
high temperature testing
PE accelerometer

Instrumentation labs, test
benches, R&D facilities

Instrumentation

Instrumentation labs,
PE/IEPE sensors

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 78
VIBRATION SENSORSWATER LEVEL SENSORSWATER LEVELSENSORS

PAGE 79
WATER LEVEL SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

WATER LEVEL SENSORSWATER LEVEL DATA LOGGERS

TruBlue Logger 555 Level,
575 Baro, 585 CTD

TruBlue Logger 255 Level

TruBlue Logger 275 Baro

Accuracy±0.05% FS TEB (TruBlue 555, 575, 585) 1%of reading or 20 µs/cm (TruBlue 585)

0.05% FS TEB

0.05% FS TEB

Range

0 - 692 ft (TruBlue 555, 585)
8 - 16 psia (TruBlue 575)
5 - 200,000µs/cm (TruBlue 585)

0 - 658 ft H2O

8 - 16 psia

Max. Over-range

2X FS (TruBlue 555, 585)
32 psia (TruBlue 575)

3X FS

3X FS

Output

RS-485, SDI - 12

RS 485, SDI - 12

RS 485, SDI-12

Data Logging
Memory8 MB

8 MB or 56 MB

8 MB or 56 MB

Operating Temp.

0°C to 50°C

0°C to 50°C

0°C to 50°C

Dimensions (mm)Ø19.0 x 390.0

Ø19.0 x 222.0

Ø19.0 x 222.0

Typical Applications

Groundwater monitoring, surface water
monitoring, oceanographic research,barometric pressure monitoring

Flood and storm monitoring, wave studies
and rapid sampling, stream and stage gaging,slug and pump test, aquifer characterization

Barometric pressure monitoring

KPSI 500, 501

KPSI 351, 353, 355

KPSI 600, 601 - Ceramic

Accuracy±0.05% FS TEB (KPSI 500) ±0.01 ft H2O(KPSI 501)

±0.10% FS TEB (KPSI 353) ±0.05% FS
TEB (KPSI 355) ±0.01 ft H2O (KPSI 351)

±0.05% FS TEB (KPSI 600) ±0.01 ft H2O(KPSI 601)

Range

10 - 230 ft (KPSI 500) 10 - 50 ft (KPSI 501)

10 - 230 ft (KPSI 353, 355) 10 - 50 ft (KPSI 351)

10 - 230 ft (KPSI 500) 10 - 50 ft (KPSI 501)

Max. Over-range

2X FS

2X FS

5X FS

OutputSDI - 12, RS-485

SDI - 12, RS-485

SDI - 12, RS-485

Operating Temp.

-20°C to 60°C

-20°C to 60°C

-20°C to 60°C

Dimensions (mm)Ø25.4 x 197.0

Ø19.0 x 243.0

Ø25.4 x 197.0

Typical Applications

Groundwater monitoring, surface water
monitoring, oceanographic research

Groundwater monitoring, surface water
monitoring, oceanographic research

Dissolved gas monitoring, trailrace egress
monitoring, ground water monitoring,
oceanographic research

DIGITAL LEVEL SENSORSSS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 80
WATER LEVEL SENSORSWATER LEVEL SENSORSDIGITAL TEMPERATURE SENSORSANALOG LEVEL SENSORS
1" BoreKPSI 380
Accuracy±0.1°C

Range-20°C to 60°C

Connection

Open port nosepiece

OutputSDI - 12, RS-485

Operating Temp.

-20°C to 60°C

Dimensions (mm)Ø19.0 x 127.0

Typical Applications

Groundwater monitoring, surface
water monitoring, storm water, dam
operations and stream gaging

KPSI 700, 710, 720

KPSI 730, 735

Accuracy±1.00%, ±0.50%,±0.25% FSO±0.10%, ±0.05% FSO

Range

Custom ranges from:
2.3 - 700 ft H2O (Vented)
10 - 700 ft H2O (Sealed)
35 - 700 ft H₂O (Absolute)

Custom ranges from:
5 - 700 ft H2O (Vented: KPSI 730)
0 - 5 ft H2O to 0 - 700 ft H2O(Sealed, Absolute: KPSI 730)
6 - 700 ft H2O (Vented KPSI 735)

Max. Over-range

2X FS

2X FS

Output4 - 20 mA, 0 - 5 VDC,
0 - 2.5 VDC, 0 - 4 VDC,
0 - 10 VDC, 1.5 - 7.5 VDC

4 - 20 mA, 0 - 5 VDC, 0 - 2.5 VDC,
0 - 4 VDC, 0 - 10 VDC,
1.5 - 7.5 VDC

Operating Temp.

-20°C to 60°C

-20°C to 60°C

Dimensions (mm)Ø25.4 x 86.6

Ø25.4 x 86.6

TypicalApplications

Groundwater monitoring,
surface water monitoring,
oceanographic research,pump control, life
stations, landfill leachate

Groundwater monitoring, surface
water monitoring, oceanographic
research, pump control, life
stations, landfill leachate

Agency Approvals

CE, WEEE, RoHS, UL and
FM (Intrinsically safe)

CE, WEEE, RoHS, UL and
FM (Intrinsically safe)

ANALOG LEVEL SENSORS
0.75” BoreKPSI 320, 330, 335, 342

KPSI 300DS

Accuracy±0.10%, ±0.05% FSO (KPSI 330, 335)
±0.25% FSO (KPSI 320)
±0.25% FS TEB (KPSI 342)

±0.50% FSO

Range

Custom ranges from:
5 - 700 ft H2O (Vented: KPSI 320, 330, 335)
10 - 700 ft H2O (Vented KPSI 342)
0 - 5 ft H2O to 0-700 ft H2O (Sealed: KPSI 330, 342)
10 - 700 ft H₂O (Sealed: KPSI 320)
35 - 700 ft H₂O (Absolute: KPSI 320, 330, 342)

Custom ranges from:700 - 6,921 ft H2OMax. Over-range

2X FS

2X FS

Output4- 20 mA, 0 - 5 VDC, 0 - 2.5 VDC, 0 - 4 VDC,
0 - 10 VDC, 1.5 - 7.5 VDC (KPSI 320, 330, 335)
4 - 20 mA (KPSI 342)

4 - 20 mA, 0 - 5 VDC, 0 - 2.5 VDC,
0 - 4 VDC, 0 - 10 VDC, 1.5 - 7.5 VDC

Operating Temp.

-20°C to 60°C (KPSI 320, 330, 335)
-20°C to 85°C (KPSI 342)

-20°C to 60°C

Dimensions (mm)Ø19.0 x 151.0

Ø19.0 x 215.0

Typical Applications

Groundwater monitoring, surface water monitoring, oceanographic
research, pump control, lift stations, landfill leachate, tailrace andforebay monitoring

Down hole, level control, pump control

Agency Approvals

CE, WEEE, RoHS, UL and FM (Intrinsically safe)(KPSI 320, 330, 335) CE, WEEE, RoHS (KPSI 342)

CE, WEEE, RoHS

PAGE 81
WATER LEVEL SENSORS

te.com/sensors

Specifications subject to change.
Dimensions for reference purpose only.

SS-TS-TE101
06/2019

WATER LEVEL SENSORSLEVEL SENSORSOEM Level Sensors

KPSI 705

KPSI 745, 750

LTA, LT

LTB, LTR

Accuracy±0.25% FSO±0.25% FSO±0.25% FSO±0.25% FSO

Options

Optional ETFE

Optional standoff (KPSI 745)

Optional lightning protection

Optional lightning protection

Range

Custom ranges from
6 - 115 ft H2O

Custom ranges from 10 - 115 H2O

0 - 1 psi up to 0 - 300 psi
Custom ranges available

0 - 11.5, 23.1, 34.6, 69.2, 115.4 ft
H2O Custom ranges available

Max. Over-range

2X FS

2X FS

2X FS

2X FS

Output4 - 20 mA, 0 - 5 VDC,
0 - 2.5 VDC, 0 - 4 VDC,
0 - 10 VDC, 1.5 - 7.5 VDC

4 - 20 mA, 0 - 5 VDC, 0 - 2.5 VDC,
0 - 4 VDC, 0 - 10 VDC, 1.5 - 7.5 VDC

4 - 20 mA

4 – 20 mA, 0 – 5 VDC,
0 – 10 VDC, 0 – 2.5 VDC,
0 – 4 VDC, 1.5 – 7.5 VDC

Operating Temp.

-20°C to 60°C

-20°C to 60°C

-20°C to 60°C

-20°C to 60°C

Dimensions (mm)Ø25.4 x 86.6

KPSI 745: Ø88.9 x 279.4 (Withstandoff) Ø88.9 x 253.3 (Without
standoff) KPSI 750: Ø104.1 x 279.4

LTA: Ø25.4 x 93.0 LT: Ø25.4 x 170.5(Dependent on fitting)

LTB: Ø104.1 x 206.5 LTR:
287.1 with overmold conduit
connection, 253.5 with glandseal conduit connection

TypicalApplications

Wastewater, lift stations,pump control, slurry tankliquid level, tank level

Wastewater, lift stations,pump control, slurry tankliquid level, tank level

Pump control, tank liquid level,
landfill leachate monitoring,
construction bypass pumping,dewatering, lift station monitoring,
submersible tank liquid level,liquid line pressure, slurry tankliquid level, wastewater

Pump control, tank liquid level,
landfill leachate monitoring,
construction bypass pumping,dewatering, lift station monitoring,
submersible tank liquid level,liquid line pressure, slurry tankliquid level, wastewater

Agency Approvals

CE, WEEE, RoHS, UL and
FM (Intrinsically safe)

CE, WEEE, RoHS, UL and
FM (Intrinsically safe)

CE, WEEE, RoHS, with optional UL,
CUL, and FM (Intrinsically safe)

CE, WEEE, RoHS, with optional
UL, CUL, and FM (Intrinsically safe)

NON-SUBMERSIBLE PRESSURE TRANSDUCERS
OEM Level Sensors

KPSI 27, 28

KPSI 30

Accuracy±0.5%, ±0.25%±0.1%

Options

IP68 submersible option

IP68 submersible optionRange

1 - 300 psi (Vented) 5 - 2000 psi (Sealed)
15 - 2,000 psi (Absolute)2 - 300 psi (Vented) 5 - 500 psi(Sealed, absolute)Max. Over-range

2X FS

2X FS

Output4-20 mA, 0-5 VDC, 0-2.5 VDC 0-4 VDC,
0-10 VDC, 1.5-7.5 VDC

4-20 mA, 0-5 VDC, 0-2.5 VDC
0-4 VDC, 0-10 VDC,
1.5-7.5 VDC

Operating Temp.

-20°C to 60°C

-20°C to 60°C

Dimensions (mm)Ø25.4 x 86.6

Ø25.4 x 86.6

Typical Applications

Line pressure monitoring, pump and liftstations, pump control, tank levelmonitoring, underwater researchLine pressure monitoring, pump and liftstations, pump control, tank levelmonitoring, underwater research

Agency Approvals

CE, WEEE, RoHS, UL and FM (Intrinsically safe)

CE, WEEE, RoHS, UL and FM (Intrinsically safe)

SS-TS-TE101
06/2019

Specifications subject to change.
Dimensions for reference purpose only.

te.com/sensors

PAGE 82
WATER LEVEL SENSORSSENSOR SOLUTIONS WORLDWIDE RESOURCES
EVERY CONNECTION COUNTSTE Connectivity is a global technology leader. Ourconnectivity and sensor solutions are essential in
today's increasingly connected world. If data, signal
or power moves through it, TE connects and senses it.

14
6
1

2
3

7

8
11

9
10

4
5
13

1

Fremont, California USA
Pressure Mfg/R&D

2

Grass Valley, California USA
Position Mfg/R&D

3

Chatsworth, California USA
Position R&D

4

Irvine, California USA
Vibration R&D

5

Hermosillo, Mexico
Pressure, Position & Automotive Mfg

6

Andover, Minnesota USA
Temperature & Optical Mfg/R&D

7

Troy, Michigan USA
Automotive R&D

8

Akron, Ohio USA
Position Mfg/R&D

9

Berwyn, Pensylvania USA
USA Headquarters
Hampton, Virginia USA

TE designs, manufactures and delivers products,
systems and solutions in over 150 countries. This
global reach enables us to work closely with ourcustomers and identify and act on local needs quickly.
By leveraging our global scale, we can deliver thehighest levels of quality, innovation and service at a
local level.

PAGE 83

te.com/sensors

10 Piezo, Force & Position Mfg/R&D
11High Point, North Carolina USA
Fluid Property & Automotive R&D

12

Winston-Salem, North Carolina USA
Fluid Property R&D

13

Empalme, Sonora, Mexico
Automotive Mfg
Montreal QC, Canada

14 Autotmotive R&D

SS-TS-TE101
06/2019

CHOOSE A PARTNER THAT'S AS GLOBAL AS YOU ARE
Connect with us today at te.com/sensors

15

16
26
22 21

19
17
18
24
20

25

23
27

34
35

28

32

3130

15

Bergen, Norway
Fluid Property R&D

16

Galway, Ireland
Temperature Mfg/R&D

17

Bensheim, Germany
Automotive R&D

18

Speyer, Germany
Automotive Mfg/R&D

19

Dortmund, Germany
Position & Temperature Mfg/R&D

31

Bevaix, Switzerland

32 Automotive R&D

21

Les Clayes-Sous-Bois, France
Force, Vibration & Position
Mfg/R&D

33 Fluid Property Mfg

22

Dreux, France
Position Mfg/R&D

Chengdu, China

28 Temperature, Position, Humidity,
Speed, Piezo, Optical Mfg

29

Shenzhen, China

29 Pressure, Force, Position, Speed,
PCBA, Fluid Property Mfg
Shanghai, China

33

30 Automotive R&D
Suzhou, China
Automotive Mfg/R&D
Kawasaki, Japan

20 Pressure Mfg/R&D

Bankok, Thailand

Seoul, Repulic of Korea

34 Automotive R&D

Kyungsan, South Korea

35 Automotive Mfg/R&D

Toulouse, France

23 Humidity Mfg/R&D
Pratteln, Switzerland

24 Speed & Position Mfg/R&D
Timisoara, Romania

25 Temperature, Speed & Fluid
Property Mfg

Oostkamp, Belgium

26 Autotmotive Mfg/R&D
Evora, Portugal

27 Automotive Mfg

SS-TS-TE101
06/2019te.com/sensors

PAGE 84

GLOSSARY OF COMMON SENSOR TERMS
Calibration

Over-range Limit

Compensated Temperature Range

Plug and Play

Testing of a sensor to confirm output is within a specified range
for particular values of the input.

The temperature range in which the sensor meets the
specifications for Thermal Zero Shift and Thermal Sensitivity Shift.

DeviceNet™

The over-range limit is the maximum input to which the sensor
can be exposed without damage.

Sensors designed for end-users who expect sensors to meetcalibration performance standards once power and signal cables
are properly connected to instrumentation.

Root Mean Square

Device level network for industrial automation.

The square root of the arithmetical mean of a set of squared
instantaneous values

Excitation

The recommended voltage with which a standard sensor should
be excited.

Full Scale Output (FSO)

Full Scale Output (FSO) is the span between the lowest range
limit and the highest range limit of the sensor. Published values
are approximate values and may vary with each sensor.

Hysteresis

Hysteresis is the difference in sensor output signal at a specificinput when applied in the increasing and then decreasing sectorsof a single cycle of short time duration at constant temperature. It
is expressed as a percentage of FSO.

Natural Frequency

Natural Frequency is the frequency at which the sensor’s
active sensing element goes into resonance and responds withmaximum movement for a specific applied input.

Non-linearity

Non-linearity is the deviation of the sensor output signal from atheoretical straight line which has been fitted to the data pointsof an actual calibration. It expresses the maximum deviation of
all data points in that calibration and is sometime expressed as apercentage of FSO, usually as a ±% error band, or % of reading.

Non-Repeatability

Non-repeatability is the deviation in sensor output signal levels
when a specific input is applied in consecutive cycles of shorttime duration under the same conditions, such as temperatureand direction of increasing or decreasing input. It can be
determined by performing two consecutive short time duration
calibration cycles and can be expressed as ±%FSO.

Operating Temperature

Sealing

Sealing is the assembly method by which the sensor is protectedfrom moisture in the surrounding environment. The most desirable
sealing method is hermetically seal. This can be achieved byjoining the individual piece parts together by soldering, welding,brazing, glassing, or other commonly accepted manufacturing
processes. Another common sealing method is epoxy seal. It
is achieved by joining the piece parts by applying adhesive or
potting compound to mitigate the incursion of moisture into thesensor assembly.

Sensitivity

The sensor’s change in output per the unit change in the physicalparameter being measured. The change may be linear or nonlinear.

Thermal Sensitivity Shift (TSS)The change in sensitivity of the sensor as a function oftemperature. It is usually expressed as a percent reading change
in sensitivity for a specified change in temperature such as±0.01%/°C and is generally linear with moderate temperaturechanges. The Thermal Sensitivity Shift can be eliminated orminimized by using sensitivity numbers determined at or near thetemperature of use.

Thermal Zero Shift (TZS)

The change in the Zero Offset as a function of temperature is the
Thermal Zero Shift. It may be expressed as either a %FSO for a
specific temperature change such as ±0.01%FSO/°C or in voltage
units such as ±0.2 mV/°C and it is not a linear function.

Total Error Band (TEB)

Typically expressed as a percentage, the TEB is the combinationof possible errors for a sensing device within its measurement
range and temperature of operation.

The temperature range within which a sensor will meet all of its
stated specifications while powered and in operation.

PAGE 85

te.com/sensors

SS-TS-TE101
06/2019

=TE

GLOSSARY OF COMMON SENSOR ABBREVIATIONS

connectMty

ABS

American Bureau of Shipping

IP

Ingress Protection

PSIS

AC

Alternating Current

ISOInternational Organizationfor Standardization

Pounds Per Square InchSealed Gage Reference

PTFE

Polytetrafluoroethylene

ITAR

International Traffic in Arms Regulations

PUDF

Public Use Data FileANSI

American National Standards Institute

ASIC

Application-Specific Integrated Circuit

ATEX

Appareils destines a etre utilises
en ATmospheres EXplosibles

kHz

Kilohertz

PWM

Pulse Width Modulation

LED

Light Emitting Diode

R&D

Research and Development

BOP

Blow Out Prevention

LIN

Local Interconnect Network

RDT&E

Research, Development, Test & Evaluation

CAN

Controller Area Network

LVD

Low Voltage Differential

RFI

Radio Frequency InterferenceCE

Communaute Europ8enne

LVDT

Linear Variable Displacement Transducers

RH

Relative Humidity

CENELEC European Committee forElectrotechnical Standardization

mA

Milliamp

RMS

Root Mean Square

MAF

Mass Air Flow

RoHS

Restriction of Hazardous SubstancesmbarMillibar

RPM

Revolutions Per Minute

MCR

Main Control Room

RTD

Resistance Temperature Detector

MEMS

Microelectromechanical Systems

RTU

Remote Terminal Unit

CSA

Canadian Standards Association

CT

Computed Tomography

cUL

Tested to Canadian Standards
by Underwriters' Laboratories

DC

Direct Current

DCS

Distributed Control System

DEF

Diesel Exhaust Fluid

DTC

Digital Temperature Compensation

ECU

Engine Control Unit

EGR

Exhaust Gas Recirculation

EMC

Electromagnetic Compatibility

EMI

Electromagnetic Interference

ESA

European Space Agency

FLS

Field Loadable Software

FM

Factory Mutual

FPGA

Field Programmable Gate ArrayFS

Full Scale

FSO

Full Scale Output

FT LBS Foot Pounds
GPS

Global Positioning System
Health Usage and Monitoring System

HUMS

mHZ

Megahertz

R VDT

Rotary Variable Differential Transformermm

Millimeter

SAE

Society of Automotive Engineering

MQS

Military Qualification Standards

SCADA Supervisory Control and Data Acquisition

MR

Magnetoresistive

SCR

Selective Catalytic ReductionmVMillivolt

SDl-12

Serial Data Interface at 1200 BaudNAV

Navigation

SMD

Surface Mount Device

NASA

National Aeronautics
and Space Administration

Sp0 2

Pulse Oximeter Oxygen Saturation

SPDT

Single Pole, Double Throw

NEMA

National Electrical
Manufacturers Association

SPI

Serial Peripheral Interface

SPST

Single Pole, Single Throw

T&M

Test & Measurement

NIST

National Institute ofStandards and Technology

NOx

Nitrogen Oxide

TDFN

Thin Duel Flats No Leads

NPT

National Pipe TaperedTE

TE Connectivity

NSF

National Science Foundation

TEB

Total Error BandNTC

Negative Temperature Coefficient

TESS

TE Sensor Solutions

OEM

Original Equipment Manufacturer

THSA

Trimmable Horizontal Stabilizer Actuators

PCB

Printed Circuit Board

TPMS

Tire Pressure Monitoring System

HVACR Heating, Ventilation,
Air Conditioning, and Refrigeration

PDF

Portable Document Format

TSYS

Temperature System Sensor

PDM

Pulse Density Modulation

UAV

Unmanned Aerial Vehicle

HVD

High-Voltage DifferentialPE

Piezoelectric

HZ

Hertz

PLCD

UL

l'C

Inter-Integrated Circuit

Permanent Magnet Linear
Displacement Sensor

USB

Universal Serial Bus

IEC

International Electrical Commission

PPS

Polyphenylene Sulfide

VAV

Variable Air Volume

IECEx

International ElectrotechnicalCommission Explosive

PSI

Pounds Per Square Inch

VDC

Volts Direct Current

PSIA

Pounds Per Square Inch-Absolute Reference

WEEE

Waste Electrical and Electronic Equipment

IEEE

Institute of Electricaland Electronics Engineers

PSID

Pounds Per Square InchDifferential Reference

IEPE

Integral Electronic Piezoelectric

PSIG

Pounds Per Square Inch-Gage Referenceuc

Microcontroller
Underwriters Laboratories© 2019 TE Connectivity. All Rights Reserved.
Android is a trademark of Google Inc.
CANopen® is a registered trademark of the CAN in Automation User's Group.
DeviceNeeM is a trademark of ODVA, Inc.
10S is a trademark or registered trademark of Cisco in the U.S. and other countries and is used under license.
Linux® is the registered trademark of Linus Torvalds in the U.S. and other countries.
Noryl® is a registered trademark of Sabic Innovative Plastics IP BV.
Pmod is a trademark of Digilent Inc. and is used under license.
Accustar, ATEXIS, DEUTSCH TruBlue, KPSI, Micro/used, ldentiCal, Krystal Bond, AST, Jaquet, TE Connectivity, TE, and the TE connectivity (logo) are trademarks of the TE Connectivity
Ltd. family of companies. Other logos, product and company names mentioned herein may be trademarks of their respective owners.
While TE has made every reasonable effort to ensure the accuracy of the information in this brochure, TE does not guarantee that it is error-free, nor does TE make any otherrepresentation, warranty or guarantee that the information is accurate, correct, reliable or current. TE reserves the right to make any adjustments to the information contained herein at
any time without notice. TE expressly disclaims all implied warranties regarding the information contained herein, including, but not limited to, any implied warranties of merchantability
or fitness for a particular purpose. The dimensions in this brochure are for reference purposes only and are subject to change without notice. Specifications are subject to change
without notice. Consult TE for the latest dimensions and design specifications.

TE-SEN-TElOl
06/2019te.com/sensors

PAGE 86