ＩＳＳＮ１３４６－７８４０

港湾空港技術研究所
資料
TECHNICAL NOTEOFTHE PORT AND AIRPORT RESEARCH INSTITUTE

No.1243

December 2011

港湾地域強震観測年報（2010）

独立行政法人

野津

厚

若井

淳

港湾空港技術研究所

Independent Administrative Institution,

Port and Airport Research Institute, Japan目

次旨 ······························································································································································

3

1. はじめに ························································································································································

4

2. 観測網と観測機器 ········································································································································

62.1

観測網 ····················································································································································

6

2.2

強震計の維持管理 ································································································································

62.3

観測地点 ················································································································································

6

2.4

強震計 ····················································································································································

7

2.5

強震計の基礎と観測小屋 ··················································································································· 10

要3. 記録の整理 ···················································································································································· 11
4. 補正と積分 ···················································································································································· 12
4.1

フィルタリングと積分の方法 ··········································································································· 124.2計器特性補正等のためのフィルタ ··································································································· 12

4.3

SMAC-B2相当フィルタ ························································································································· 12

4.4

積分用のハイパスフィルタ ··············································································································· 13

4.5

解析のアウトプット ···························································································································· 14

5. 2010年6月13日福島県沖の地震（M6.2）の記録について ···································································· 14
6. まとめ ···························································································································································· 18
謝辞 ······································································································································································· 18
参考文献 ······························································································································································ 18
強震観測担当者（2010年1月～12月） ··········································································································· 21-1-Annual Report on Strong-Motion Earthquake
Records in Japanese Ports (2010)
Atsushi NOZU*
Atsushi WAKAI**

Synopsis
Since 1962, strong ground motions and earthquake responses of structures have been observed in
the major ports in Japan. By the end of December 2010, 9411 accelerograms had been accumulated
and analyzed at the Port and Airport Research Institute.
The strong-motion earthquake observation network in Japanese ports consists of 119
strong-motion accelerographs installed at 61 ports as of December 2010. Sixty-nine accelerographs
out of 119 are installed on ground surface, 36 accelerographs are in ground by using bore-hole andthe rest 14 are on structures such as quay walls. Currently two types of accelerographs are used in
the network, namely, the ERS accelerograph and the SMAC-MDU accelerograph, both of which areof an electrical type equipped with a digital recorder.
This report presents the results of observation and preliminary analysis of records obtained in2010. ASCII data of the reproduced accelerograms for all the records in 2010 are available on theattached CD in the user-friendly CSV format. Also, computer plots of reproduced accelerograms areavailable on the CD for the records with peak accelerations exceeding 20 Gal. For the records withpeak accelerations exceeding 50 Gal, computer plots of reproduced accelerograms, integrated
velocities and displacements, Fourier spectra and response spectra are available on the CD.
Key Words: earthquake, port, strong-motion earthquake observation, digitized acceleration record,
Fourier spectra* Head of Engineering Seismology Division, Geotechnical and Structural Engineering Department** Postdoctoral Fellow, Engineering Seismology Division, Geotechnical and Structural Engineering Department
3-1-1 Nagase, Yokosuka, 239-0826 Japan
Phone：+81-46-844-5058
Fax：+81-46-844-0839
e-mail:nozu@pari.go.jp-2-港湾地域強震観測年報（2010）
野津

厚*・若井

要

淳**

旨

1962 年より実施されている港湾地域強震観測においては，2010 年末現在，9411 の強震記録が蓄積されるに至っている．このうち 2010 年には 296 の強震記録が得られている．強震計の台数としては，2010 年末現在，119 台の強震計が 61 港に設置されており，設置状況としては，69 台が地表に，
36 台が地中に，14 台が構造物上に設置されている．観測網を構成する強震計は ERS-F,-G,-GV 型強震計と SMAC-MDU 型強震計の 2 種類であり，いずれも電気式デジタル記録方式である．
本資料は 2010 年に港湾地域強震観測網で得られた記録について報告するものである．得られた記録は港湾毎に分類され，地震諸元・記録番号・最大加速度等とともに付録 CD の強震観測表にまとめられている．強震観測表の地震諸元は気象庁の地震・火山月報（カタログ編）に基づいている．
また，強震記録の記録番号は各観測地点から記録が送られてきた順番に付けられており，記録の分類として，頭文字「F」で始まる記録は ERS-F,-G,-GV 型強震計で得られた記録，頭文字「U」で始まる記録は SMAC-MDU 型強震計で得られた記録をそれぞれ示している．強震観測表にまとめられているすべての記録のデジタルデータを付録 CD に CSV 形式で収録している．最大加速度が 20Gal を越える記録については加速度波形を画像ファイルの形で付録 CD に収録している．また，最大加速度が 50Gal を越える比較的大きな記録については，加速度波形に加え，速度波形，変位波形，フーリエスペクトル，応答スペクトルも画像ファイルの形で付録 CD に収録している．
2010 年の観測は次に示す諸機関の協力の下に実施された．
国土交通省港湾局
東京都港湾局

国土交通省地方整備局

静岡県港湾総室

国土交通省北海道開発局

大阪市港湾局

宮崎県港湾課

内閣府沖縄総合事務局

国土技術政策総合研究所

強震観測は，各観測地点での強震観測担当者の努力に負うところが非常に大きい．担当者各位に敬意と謝意を表すとともに，各観測地点で実際に観測に携わった方々の氏名を巻末に掲載する．
キーワード：地震，港湾地域強震観測，数値化加速度記録，フーリエスペクトル* 地盤・構造部 地震防災研究領域 地震動研究チームリーダー** 地盤・構造部 地震防災研究領域 地震動研究チーム 特別研究員〒239-0826 横須賀市長瀬3-1-1 港湾空港技術研究所
電話：046-844-5058 Fax：046-844-0839
e-mail:nozu@pari.go.jp-3-3），4）

1. はじめに．強震観測により地震動の卓越周期の解明が進めば，構造物の固有周期と地震動の卓越周期が一致しないよう
港湾地域強震観測は当所の前身である運輸省港湾技術に工夫することも可能になるので，地震による被害の軽研究所が中心となり 1962 年に開始され，国の機関や地方減に資するものと期待される．

自治体が参画して実施されてきた．観測網は年々拡大し，（3）強震観測の第三の目的は，良質な（SN 比の高い）

2010 年末の時点では全国 61 の港に 119 台の強震計が設

中小地震の記録を取得して，将来の大地震の揺れの予測

置されている（図-1）．この観測網により，2010 年末まに活用することである．上述のように地震動の性質は地でに 9411 の加速度記録が得られている．このうち 2010

点毎に異なるのであるから，構造物の耐震設計には地点

年には 296 の記録が得られている．参考までに，2001 年毎の特性を反映した入力地震動を用いることが望ましい．

以降の強震計台数と記録数の推移を表-1 に示す．なお，

各港湾で得られた中小地震の記録には，港湾毎の地下構

図-1 では，ERS-F 型強震計の設置されている地点を濃い造の情報が含まれており，この情報を生かすことにより，緑で，SMAC-MDU 型強震計の設置されている地点を薄将来の大地震による揺れの予測に役立てることができる．

い緑で示している．強震計の機種については 2.を参照さ

その一例として，2000 年鳥取県西部地震の余震の際に境れたい．

港とその周辺の複数の地点（図-3）で取得された記録を

港湾地域強震観測の目的は極めて多岐にわたるが，こ

経験的グリーン関数法と呼ばれる方法で多数重ね合わせこでは代表的な三つの目的を説明する．

ることにより，鳥取県西部地震の本震地動をシミュレー（1）強震観測の第一の目的は，港湾に被害をもたらすよ

ションした結果を図-4 に示すが，各地点の地震動の振幅，

うな大地震が発生した場合に，その揺れを記録して，こ周期，継続時間などの特徴をかなり良好に再現できていれを被害メカニズムの解明や適切な復旧工法の選定に利ることがわかる．ここで用いた手法の詳細については文用することである．1995 年兵庫県南部地震で被災した神献 5）を参照していただきたい．

戸港の岸壁の被害原因の解明には，神戸港で取得された
強震記録が活用された

港湾地域強震観測で得られた記録は，後に説明するよ

1），2）．また，本資料の対象期間外

うな処理と解析を経た後に，強震観測年報として公表さであるが，2011 年東日本大震災の際に取得された強震記

れてきた．まず，1963 年から 1975 年の記録が強震観測

録も，防波堤等の被害原因の解明に活用されている．

年報として公表された 6)-16)．1963 年から 1975 年の年報

（2）強震観測の第二の目的は，地点毎に異なる地震動のには鉛直成分が含まれていなかったので，それらは別冊

特性を解明することである．図-2 では八戸港と関西国際

17)

空港で得られた強震記録のフーリエスペクトルを比較しからは新しいデータ処理の方法を取り入れ，計器補正をている．八戸港では 1968 年十勝沖地震と 1994 年三陸は施した加速度波形とそれを積分した速度波形と変位波形，

るか沖地震の強震記録が得られているが，26 年の時を隔

フーリエスペクトルや応答スペクトルを年報で報告するてて発生した二つの大地震で，いずれも周期 2.5 秒（周ようになった 18)-46)．また，通常の年報とは別に，被害地

波数 0.4Hz）の成分が卓越している．一方，関西国際空震の記録だけを集めた資料も刊行されている 47)-58)．強震

港では 1995 年兵庫県南部地震と 2000 年鳥取県西部地震

観測年報の変遷を表-2 に示す．

として公表された．1976 年と 1977 年の強震観測年報の記録が得られているが，いずれも周期 5 秒（周波数

昨今のように公的機関の組織再編が進む中では，強震

0.2Hz）の成分が卓越している．このように，場所により記録のデジタルデータを研究室内で保管することが最も地震動の特性が異なるのは，地震基盤上に存在する堆積

確実な保管方法であるとはもはや言えなくなってきてい層の地震動への影響（サイト特性）が場所毎に異なるたる．デジタルデータを保管する場所としてはウェブ・サめである．平成 19 年 4 月に改訂された新しい港湾の施設

ーバも考えられるが，ウェブ・サーバのサービスが停止の技術上の基準では，設計用入力地震動の評価を行う際，すれば，それと同時にデータが失われる可能性がある．

対象施設設置位置もしくはその周辺における地震動の実

しかるべき出版物に電子媒体を添付して，その中にデジ

測値に基づいてサイト特性の評価を行うことが求められ

タルデータを収めておくことが，強震記録のデジタルデるようになった．従って，設計実務において強震観測のータを後世に残すための最も確実な手段であろう．電子

果たす役割の重要性は今後一層高まるものと予想される．

媒体にも種々のものがあるが，容量・価格・普及の度合

このことに関連して，全国の港湾等の強震観測地点においなどから CD が最適と判断される．このような考えか

けるサイト増幅特性の評価結果をとりまとめた資料が刊

ら 1999 年以降の強震観測年報 37)-46)には，デジタルデー

行されているので，参考にしていただければ幸いである

タを収録した CD を添付している．

-4-

図-3 境港およびその周辺の強震観測地点

図-1 港湾地域強震観測網
図-4 経験的グリーン関数法による 2000 年鳥取県西部
地震の本震地動の再現 5）
表-1 強震計台数と記録数の推移（台数は年末の数字）
年 港湾数 強震計台数 地震数 記録数
2001
60
97
67
207
2002
60
98
50
140
2003
60
104
109
394
2004
60
110
113
468
2005
60
111
98
434
2006
61
113
88
276
2007
61
119
173
520
2008
61
119
96
417
2009
61
119
131
488
2010
61
119
99
296

表-2 強震観測年報の変遷

年1963～1975

強震観測年報の内容
CD
加速度波形，フーリエスペクトル，応 無
答スペクトル．鉛直成分は別冊として公表．
1976～1998
計器補正を施した加速度波形とそ 無れを積分した速度波形と変位波形，
フーリエスペクトル，応答スペクト
1999～本資料 同上有

図-2 八戸港と関西国際空港の地震動特性の比較-5-

本資料の付録 CD には，2010 年に取得されたすべての答を記録するものである．構造物の地震応答を記録する

加速度記録のデジタルデータが CSV 形式で収められてための観測点のそばには必ず地盤の加速度を測定するたいる．また，20Gal を越える記録については，加速度記

めの観測点が設けられている．

録を図化したものが収められている．さらに，50Gal を

各観測地点では，原則として，強震計の水平成分の一越える記録については，加速度記録と，これを積分してつは真北を向いている．ただし，例外として，構造物の得た速度波形，変位波形，フーリエスペクトル，さらに応答を記録する観測点と，これに付随して地盤加速度を

応答スペクトルを図化したものが収められている．

記録する観測点では，強震計が構造物と平行に設置されている場合もある．岸壁や桟橋など多くの港湾構造物は2. 観測網と観測機器

2 次元的な形状をしており，構造物の法線方向に垂直な成分と平行な成分を計測する方が解析上便利であるため，

2.1

そのようにしている．強震計の 2 つの水平成分が真の南

観測網

港湾地域強震観測網は日本全国の海岸線をカバーして北方向と東西方向に一致しない場合には，方向成分を表

おり，2010 年末の時点では 119 の強震計が 61 の港に設すために偏角（単位は度）が使用される．例えば N10E

置されている．2010 年末の時点で強震計の設置されていは真北から東に 10°回転した方向を意味する．

る港湾は図-1 および付録 CD の地図に示すとおりである．

観測網のそれぞれの観測地点には観測地点名が付けら付録 CD の地図上の港名をクリックすれば，強震計の機れている．観測地点名は位置と強震計の型と設置条件の種と設置条件が表示される．観測地点の土質柱状図等も組み合わせである（図-5）．例えば北海道の釧路港の観測

付録 CD に収められており，地図からたどることができ点は「釧路-G」および「釧路-GB」と名付けられている．

る．観測地点のさらなる詳細情報については観測地点資「釧路」は観測地点の場所を意味している．港湾事務所

59)-64)を参照されたい．2010 年末の時点では，119 の観に設置されている観測点では「小名浜事」のように「事」

測地点のうち 77 地点に ERS 型強震計が，残りの 42 地点という文字をつけている．添え字の「G」は強震計の型

料に SMAC-MDU 型強震計が設置されている．これらはいを表しており釧路港には ERS-G 型強震計が設置されてずれもデジタル式の強震計である．以前に主力であった

いることを意味している．SMAC-B2 型強震計には添え

アナログ式の SMAC-B2 型強震計は姿を消している．

字「S」が，ERS-B，C，D 型の強震計には添え字「M」
が，ERS-F 型強震計には添え字「F」が，ERS-G/GV 型強2.2

強震計の維持管理

震計には添え字「G」が，SMAC-MDU 型強震計には添え

強震計の維持管理は当所と港湾地域強震観測の他の参字「U」がそれぞれ対応する．
「釧路-GB」で添え字「G」

画機関との緊密な協力の下に実地されている．現在，の後ろの「B」は設置条件を示し，ボアホールを使用しERS-G/GV 型強震計と SMAC-MDU 型強震計のほとんどて地中に強震計を設置していることを意味している．例

は電話回線に接続され，観測点と当所との間でオンライ

えば「釧路-G」のように設置条件を表す添え字がない場

ンデータ転送が可能な状況となっている．このオンライ合は，その強震計が地表に設置されていることを意味しンシステムは 2010 年末の時点では 119 の強震観測地点のている．添え字「R」がついている場合は強震計が構造

うち 111 地点に設置されている．オンラインシステムは，物上に設置されていることを意味する．
「神戸」と「六データ収集だけでなく強震計の健全度把握にも用いられており，強震計の点検等に係る現場の負担を軽減することに寄与している．2001 年 12 月には初の試みとして携
帯電話を利用した強震計のオンライン化が大船渡港で実現した．これにより，これまで電話線を引くことができないためにオンライン化が難しかった埠頭上のような場
所でも，強震計のオンライン化が進むようになってきた．
2.3

観測地点

観測網には 3 種類の観測地点がある．1 つ目は地表面の加速度を記録するもの，2 つ目はボアホールを使用し地中の加速度を記録するもの，3 つ目は構造物の地震応図-5 観測地点名のルール-6-

甲」の観測点は地中の異なる深さに強震計が設置されている．この場合，添え字「B」の後ろに強震計の深度を表す「40」等の数字をつけている（単位 m）．また，「清
水日の出-UC1」と「清水日の出-UC2」の「C」は，強震計が CDM 改良地盤に設置されていることを示している．
2.4

強震計

(1) ERS-F 型強震計
ERS-F 型強震計は不揮発性磁気バブルメモリを備えたデジタルタイプの強震計である．記録システムのメイン
ユニットは図-6 に示されており，1MB の不揮発性磁気バブルメモリが 4 つと制御装置からなり，その寸法は240×240×35mm，質量は約 1kg である．記録システムには 2 つのユニットが設置できる．時間信号はこのバブル
メモリの 1 つのチャンネルに記録され，加速度 3 成分に時間信号を加えた 4 チャンネルの記録時間はユニットが2 つの場合には約 40 分である．

図-7 一体型の ERS-F 型強震計

ERS-F 型強震計の中には，もともと標準的な型式として図-7 に示す一体型のものがあり，換振器と磁気バブル
メモリを含む記録器を 1 つの箱に収納していた．その後，換振器と記録器が分かれた分離タイプのものが開発された．図-8 に示されているものは，ボアホールを使用して地中に設置し，地中ないし基盤における地震動を観測するための換振器である．図-9 に示す換振器は構造物に取り付けるタイプのものである．磁気バブルメモリを含む
ERS-F 型強震計の記録器を前から見た様子を図-10 に示している．2010 年末現在，稼働中の ERS-F 型強震計は 3台であり，これらは分離タイプである．
ERS-F 型強震計のブロックダイアグラムを図-11 に示
す．ERS-F 型強震計の仕様を表-3 に示す．換振器の周波数特性を図-12 に示す．収録時間は最短 1 分で，加速度

図-8

図-9
図-6

ERS-F 型強震計の記録システム-7-

ERS-F, G, GV 型強震計の地中設置用換振器

ERS-F, G, GV 型強震計の構造物取り付け用換振器

表-3

ERS-F, G, GV 型強震計の仕様

総合特性(-F)2G
計測可能加速度 (-G, -GV)0.008G～2G の範囲で設定
可能
周波数範囲

0.01Hz-35Hz

ダイナミックレ
86dB over
ンジ
換振器
成分

水平 2 成分と上下成分(-F) 2G
計測可能加速度 (-G, -GV)0.008G～2G の範囲で設定
可能
感度

10-5

形式

(-F and -G)フォースバランス(-GV)速度帰還型

図-10

ERS-F 型強震計記録システムの外観

フィルタ
ハイパスフィル
0.007Hz -6dB/octave
タ
ローパスフィル
35Hz -18dB/octave
タA/D 変換
解像度

16bit

変換速度

100Hz

遅延メモリ： 10s
時計: 内蔵時計の精度は 1/100s，1 時間毎にラジオの時報により校正
図-11

スタータ: トリガーレベルは計測可能加速度の 0.5，1
または 2％
記録器

ERS-F 型強震計のブロックダイアグラム

レベルのモニタリングにより 1 分刻みで最高 10 分まで延チャンネル数(-F) 4-10 チャンネル (うち 1 チャン
ネルは時刻)(-G, -GV)3-12 チャンネル

メモリサイズ(-F) バブルメモリ: 1 Mbytes×2(-G, -GV) IC カードメモリ: 2 Mbytes

収録時間

1 記録あたり 1-10min

最大記録時間(-F) 40 min/4ch(-G, -GV) 52 min/3ch
加速度が大きいものを優先的に保存

関連情報

観測地点, 記録数,記録開始時刻, 各
成分の最大値

長することができる．トリガーの 40 秒後，もしくは収録を延長した時刻の 40 秒後における加速度レベルがトリ
ガーレベルよりも高い場合に記録時間が延長される．記録システムのメインユニットは最大で 3 成分 60 秒の地震
動を 20 個記録することができる．連続して地震が発生し，記録器の記憶容量を越えそうになった場合，最大加速度の大きな記録が残るようになっている．しかし，例外として，180 秒を越える記録は最初のものを優先する．
(2) ERS-G/GV 型強震計
ERS-G/GV 型強震計は ERS-F 型強震計を改良したものである．強震計の換振器は ERS-F 型強震計のものとほぼ

校正：総合的な校正が可能同じである．ERS-G/GV 型強震計では図-13 に示すようバックアップ電源: 充電後 2 時間

な IC カードが記録に用いられ，また，制御システムもや

容器：アルミニウム箱，防水や改良されている．IC カードのメモリサイズは 2MB で

サ イ ズ ： (-F) 54(L)×54(W)×38(H)cm (-G, -GV)
54(L)×54(W)×33(H)cm

ある．加速度 3 成分の記録時間は 1 つのカードで約 52分である．加速度のトリガーレベルは様々な段階に設定

-8-

図-12

図-13

ERS-F, G 型強震計の周波数特性

図-14

ERS-GV 型強震計の周波数特性

ERS-G/GV 型強震計の IC カードメモリ

できる．強震計の計測可能最大加速度は 2G で最大加速度のレベル（感度）は 0.008G から 2G の間で適切な値に設定できる．2010 年末の時点で ERS-G/GV 型強震計の総図-15

ERS-G/GV 型強震計の外観

数は 74 である．
ERS-G/GV 型強震計の仕様を表-3 に ERS-F 型と共に示性は ERS-F 型強震計と同じであり，図-12 に示すとおり

す．ERS-G 型の換振器はフォースバランス式であり，である．ERS-GV 型強震計の周波数特性は ERS-F,G 型とERS-GV 型の換振器は速度帰還型である．ERS-GV 型強

異なっており，これを図-14 に示す．換振器と記録器を 1

震計は，雷に対する安全性から，地中観測点と，これにつの箱に収納した標準的な ERS-G/GV 型強震計の外観を付随する地表観測点（例えば「釧路-GB」と「釧路-G」）

図-15 に示す．

で用いられるようになった．ERS-G 型強震計の周波数特

非オンライン化地点の場合，地震を記録後，記録シス

-9-

表-4

SMAC-MDU 型強震計の仕様（港湾地域強震観測での設定）
計測
計測範囲±2097Gal

振動数範囲

DC～30Hz

ダイナミック114dB
レンジ
A/D 変換解像度

24bit

サンプリング
100Hz
周波数
換振器
成分

水平 2 成分と上下成分

感度

3V/G

形式

フォースバランス

フィルタ

図-16

ハイパスフィ
0.02Hzルタ

SMAC-MDU 型強震計の外観

コネクタ部分にエリミネータを取り付けたバブルメモリ，

ローパスフィ30Hz
ルタ

ERS-G/GV 型の場合には IC カード）はケースに入れて港湾空港技術研究所に郵送される．この記録媒体はコンピ

時計
水晶精度

誤差±0.5ppm 以下

時刻修正

GPS による時刻修正

ュータに接続された再生器にセットされ，地震動の加速度時刻歴が再生される．トリガー時の絶対時刻もまた強震計内蔵の時計の記録から得られる．表-3 に示すように

記録

記録システムは 10 秒のデジタル遅延メモリを有する．も起動条件

指定 3 成分の論理演算組み合わせ

記録開始

正秒記録開始

記録停止

停止設定レベルを下回って設定時間
後に自動停止最大記録時間600 s/3ch
加速度が大きいものを優先的に保存

ルタイプの強震計である．表-4 に SMAC-MDU 型強震計

メモリ

IC カードメモリ: 10Mbyteの仕様を示す．強震計の外観と理論的な周波数特性をそれぞれ図-16 と図-17 に示す．SMAC-MDU 型強震計は

記録内容

加速度波形データ，起動時刻
設定値，機器状態，最大加速度値
SI 値，計測震度値

しも地震の初期微動の前に記録が始まっている場合には，初期微動に先行する記録の一部が削除される．
(3) SMAC-MDU 型強震計
SMAC-MDU 型強震計は 2001 年から導入されたデジタ

2010 年末の時点で全国の港湾に 42 台設置されている．
これらはすべて当所と電話回線で接続されており，強震電源計の健全度把握とデータ収集はオンラインで実施される．

使用電源

AC100V

50/60Hz

内蔵充電池

DC12V
バックアップ:3 時間以上

このことは，強震計の点検等に係る現場の負担を軽減することに寄与している．何らかの事情でオンラインでのデータ収集が不可能となった場合には，データを収録し外形寸法・質量た媒体を郵送する．

サイズ(据置
398W×418D×210Hmm，約 24kg型)

2.5 強震計の基礎と観測小屋
港湾地域強震観測で強震計を地表に設置する場合，強テムのメインユニット（メモリ）が取り出され，次の地

震観測小屋に設置する場合とハンドホール内に設置する震を記録するため，他のメモリに置き換えられる．取り場合がある．このうち強震観測小屋に設置する場合，図出されたメインユニット（つまり，ERS-F 型の場合には-18 に示すような鉄筋コンクリート製の簡単な基礎を使- 10 -

図-19

観測小屋の一例(小名浜事-G,GB)用することが標準となっている．図-18 に示される厚さ300mm の空洞は，基礎のみかけの密度を周囲の土の密度と等しくすることにより，基礎が存在することによる記録への影響を最小限とするために設けられている．基礎は強震計を収納する小屋とは構造的に離れている必要がある．通常，強震計の基礎を支持するのに杭は使用されない．しかし，観測地点の地盤が非常に軟弱な粘土地盤
図-17

SMAC-MDU 型強震計の周波数特性

やゆるい砂地盤である場合には，コンクリートや木製の杭が使用されることもある．例えば新潟港の観測点では基礎を杭で支持している．強震観測小屋そのものの構造には特に制限はないが、耐震性に優れたものである必要がある．強震観測小屋の一例として，小名浜港の強震観
測小屋を図-19 に示す．

3. 記録の整理
当所に到着したすべての強震記録は以下に述べる手順
で整理・保管される．まず，各々の強震記録は，研究室に到着した順に番号がつけられる．ERS-F,G,GV 型強震計の記録には「F」で始まる番号，SMAC-MDU 型強震計の記録には「U」で始まる番号が付けられる．次に，各々の強震記録に対応する地震を確認する．記録に対応する
地震の震源情報としては，気象庁地震・火山月報（防災編）65）のものを仮に採用しておき，強震観測年報の刊行までには気象庁地震・火山月報（カタログ編）66）のものに置き換える．ただし，今回の強震観測年報では，10 月
以降については，カタログ編の入手時期の関係で防災編の震源情報を採用している．付録 CD の強震記録一覧のマグニチュードは気象庁マグニチュードである．
図-18

強震計基礎の一例- 11 -

図-21

SMAC-B2 相当フィルタ

ジナル加速度波形）の外側にゼロを付けたものに対して
FFT を適用する．ゼロ部分の長さ L(s)は次式により定め
る．

L > max [2 / 3T,10.0](1)

ここに T(s)はオリジナル加速度波形の長さである．この
条件は，後述する積分用のハイパスフィルタのインパル
ス応答を考慮して定めたものである．
4.2 計器特性補正等のためのフィルタ
計器特性補正としては，図-12，図-14 および図-17 の下段に示す位相特性に関する補正を行う．振幅特性に関
図-20

しては，高周波成分を次式に示すフィルタにより取り除補正と積分の手順

く．

4. 補正と積分
Af f得られた記録に対して，まずゼロ線補正が行われる．
次に，定められた方法

67), 68)={1| f | ≤ f1

1 [cos (π( f – f )/( f – f ))+1]
1
2
1
2
0

f1≤| f | ≤ f2
f2≤| f |(2)

に従い，計器特性補正のた
ここに f1=25Hz， f2=40Hz である．

めのフィルタリング，高周波成分と低周波成分の補正のためのフィルタリング，積分，応答スペクトルとフーリ

SMAC-B2 相当フィルタ

エスペクトルの計算などを実施する．処理の手順を図-20

4.3に示す．この手順は 1976 年以降の記録に適用されてきて港湾地域強震観測で長く用いられてきた SMAC-B2 型

いる．図-20 においてオリジナル加速度波形とはゼロ線

強震計の周波数特性は ERS 型強震計及び SMAC-MDU 型

補正のみを行った波形，補正加速度波形とは，計器特性

強震計とは異なるので，これらの互いに異なる機種の強に関する補正を行った上で，SN 比の十分でない低周波

震計による記録の相互比較を可能にするため，ERS 型強成分をハイパスフィルタにより削除した波形，SMAC-B2

震計及び SMAC-MDU 型強震計の記録に対しては次式に相当加速度波形とは，港湾地域強震観測で長く用いられ示すフィルタを適用する．

てきた SMAC-B2 型強震計による記録と比較可能なようS (f) =に周波数成分を調整した波形のことである．

1–

f
fs

2

(3)

1
+ 2h sfi
fs

4.1 フィルタリングと積分の方法

ここに fs=1/0.14=7.1Hz，hs=1.0 である．

フィルタリングと積分は周波数領域で実施する．リン

このフィルタは図-21 に示されるとおり，SMAC-B2 型

ク効果を避けるため，ゼロ線補正後の加速度波形（オリ

強震計と同じ周波数特性を有する．このフィルタは，ERS- 12 -

速度よりも小さいが，この処理を施して得られる加速度はすべて互いに比較可能である．
4.4 積分用のハイパスフィルタ
すべての周波数に対して SN 比を一定以上に保つと同時に物理的に意味のある信号はできるだけ消さないようにするという観点からは，積分用のハイパスフィルタの遮断周波数が加速度波形の周波数特性に応じて変化することが望ましい．一方，積分後に得られる速度波形や変位波形を異なる記録について相互に比較するという観点
からは，遮断周波数が一定値であるほうが都合がよい．
強震記録が様々な立場で応用されることを考慮し，ここ
図-22では，速度波形や変位波形を得るための積分方法として

固定フィルタと二回積分の合成

二通りの方法を用いる．一つは固定フィルタを用いる方法であり，もう一つはパラメタ付きフィルタを用いる方法である．
(1) 固定フィルタ
固定フィルタは次式により定義される．

H 1( f ) =

1

f
1– 0
f

2

f
– 2h 0 i
f

(4)

1

⋅

f1
f

1+

2

ここに f0=1/6(Hz), h=0.552，f1=0.1(Hz)である．
このフィルタは，得られる変位波形が気象庁の 1 倍強震計（T=6 秒，h=0.552）の変位波形と比較できるように設定されたものである．フィルタの遮断周波数(3 dB
図-23

down)は 0.154Hz である．なお，ここで言う「フィルタのパラメタ付きフィルタと二回積分

遮断周波数(3 dB down)」とは，フィルタの値がフラット部の 1/100.3 倍まで低下する周波数のことである．固定フィルタを図-22 に示す．
(2) パラメタ付きフィルタ
このフィルタは次式により定義される．

f
H 2( f ) = 1 – exp –
fc

2

2

(5)

このフィルタの遮断周波数は 1.36fC(3 dB down)である．
パラメタ付きフィルタを図-23 および図-24 に示す．パラ
メタ fC は，低周波側での SN 比が良好な記録に対しては小さな値，そうでない記録に対しては大きな値とする．
図-24

具体的には，以下に述べるσと E が等しくなるように fC

パラメタ付きフィルタと二回積分の合成を設定する．まず，σは次式により定義される．
型強震計及び SMAC-MDU 型強震計の記録に計器特性補σ2 = 1

T

正のためのフィルタを適用した後に適用する．このフィ

∞
-∞

X f

2

1 - exp - fT 2

4

1 - H2 f

2df

ルタを作用させることによって得られた加速度は(6)「SMAC-B2 相当加速度」と呼ばれる．ここで得られる

ここに T はオリジナル加速度波形の長さ，X(f)はオリジ

加速度は強震計で測定しようとしている地盤や構造物のナル加速度波形のフーリエ変換である．式(6)は，図-25

本来の加速度とは異なっており，その最大値は本来の加に示すように，オリジナル加速度波形から計算したフー- 13 -

している．
ここで用いる二つのハイパスフィルタはいずれもなだ
らかな周波数特性を持っており，遮断周波数の前後の成分が時間領域に変換したときに不自然な形をとって現れるのを防ぐことができる．
4.5 解析のアウトプット(1) 加速度，速度，変位各種の加速度波形と速度波形，変位波形が付録 CD に収められている．先に述べたように，オリジナル加速度
波形とはゼロ線補正のみを行った波形，補正加速度波形とは，計器特性に関する補正を行った上で，SN 比の十分でない低周波成分をハイパスフィルタにより削除した
波形，SMAC-B2 相当加速度波形とは，港湾地域強震観
測で長く用いられてきた SMAC-B2 型強震計による記録と比較可能なように周波数成分を調整した波形のことである．速度と変位は，固定フィルタによるものとパラメ
タ付きフィルタによるものの双方を提示している．
(2) フーリエスペクトル
フーリエスペクトルは補正加速度から FFT により計算される．
(3) 応答スペクトル

図-25 式(6)の意味するもの

応答スペクトルは補正加速度から計算される．絶対加速度比，絶対加速度，相対速度，相対変位の 4 種類の応

リエスペクトルの自乗を 1/T～fC の範囲で積分した値を2示している．これが所定の値（E ）となるように fC が設

答が 0%，1%，5%の 3 種類の 1 自由度系に対して計算さ定されるので，低周波成分の振幅が大きい記録，すなわ

れる．

ち低周波側での SN 比が良好な記録に対しては小さめの

5. 2010 年 6 月 13 日福島県沖の地震（M6.2）の記

fC が設定され，そうでない記録に対しては大きめの fC が録について設定される．1/T 以下の周波数成分については，fC の決定の際には考慮に入れない．
なお，E は以下に示す値である．

例年，強震観測年報では，その年に発生した代表的な

ERS-F,G 型強震計

地震の記録について一章を設けて紹介してきている．

E = ( p×0.001)×0.02236 (Gal)2010 年の 1 年間に最も大きな最大加速度をもたらした地
15

ここに p は ERS-F,G 型強震計の感度(Gal/2 )震は 6 月 13 日に福島県沖で発生した地震（M6.2）であERS-GV 型強震計

るので，ここではこの地震の記録について紹介する．

E = ( p×0.001)×0.07071 (Gal)

この地震の震央を，強震記録の取得された港湾ととも

ここに p は ERS-GV 型強震計の感度(Gal/215)に図-26 に示す．この地震では東北地方から東海地方にかけて 6 つの港湾で 16 の強震記録が得られた（図-26）．

SMAC-MDU 型強震計
E = ( p×0.001)×0.00707(Gal)

得られた加速度波形の例を図-27～28 に示す．ここでは
23

ここに p は SMAC-MDU 型強震計の感度(Gal/2 )比較的震源に近い福島県の 2 つの港湾で得られた加速度

ERS-F,G,GV 型強震計及び SMAC-MDU 型強震計の E波形を示している．図の右上に示された「相馬-G」など

値は，非地震時に取得した記録のノイズレベルを調べての文字は観測地点名を示している．相馬港で観測された定めたものである．ERS-F,G 型（フォースフィードバッ

最大加速度は 202Gal となっており，これが 2010 年に港ク）と ERS-GV 型（速度帰還型）の低周波側のノイズレ

湾地域強震観測で得られた最も最大加速度の大きい記録

ベルには差が認められたので，両者の E 値は異なる値とであった．

- 14 -

図-26

図-27

2010 年 6 月 13 日福島県沖の地震（M6.2）の震央（×）と記録の得られた港湾の位置（▲）

相馬-G におけるオリジナル加速度波形

図-28- 15 -

小名浜事-G におけるオリジナル加速度波形

図-292010 年 6 月 13 日福島県沖の地震における K-NET

図-32

相馬と相馬-G のフーリエスペクトル図-30

2003 年 5 月 26 日宮城県北部の地震における K-NET

の比

図-33

相馬と相馬-G のフーリエスペクトル図-31

2005 年 8 月 16 日宮城県東方沖の地震におけるK-NET 相馬に対する相馬-G のフーリエスペクトル2010 年 6 月 13 日福島県沖の地震における K-NET
勿来と小名浜事-G のフーリエスペクトル

図-34

K-NET 相馬と相馬-G のフーリエスペクトル

2000 年 7 月 21 日茨城県東方沖の地震におけるK-NET 勿来と小名浜事-G のフーリエスペクトル- 16 -

て，K-NET 相馬に対する相馬-G のフーリエスペクトルの比を求めると図-32 のようになり，やはり 1.5Hz 付近では
相馬-G の方がスペクトルがやや小さい傾向が表れている．
この傾向が地震を問わず表れていることから，これはK-NET 相馬と相馬-G におけるサイト特性（堆積層の存在が地震動に及ぼす影響）の相違によるものと考えられる．
図-33 は小名浜事-G で得られた記録（地表）のフーリエスペクトルを最寄りの K-NET69)観測点である K-NET 勿来（FKS012；いわき市錦町）における記録（地表）のフーリ
エスペクトルと比較したものである．この図から，小名浜事-G のスペクトルは 1.8Hz 付近では K-NET 勿来より小さめであることがわかる．このような傾向が過去の地震にも表れているか検討するために，
図-34～図-35 では 2000 年 7
図-35

図-36

2008 年 5 月 8 日茨城県東方はるか沖の地震におけ

月 21 日茨城県東方沖の地震（M6.4）および 2008 年 5 月 8

る K-NET 勿来と小名浜事-G のフーリエスペクトル日茨城県東方はるか沖の地震（M7.0）による K-NET 勿来

K-NET 勿来に対する小名浜事-G のフーリエスペク
トルの比

図-37

地表と地中のスペクトル比（相馬-G／相馬-GB）

図-38

地表と地中のスペクトル比（小名浜事-G／小名浜

図-29 は相馬-G で得られた記録（地表）のフーリエスペ
ク トルを最 寄りの K-NET69) 観 測点である K-NET 相 馬（FKS001；相馬市中村）における記録（地表）のフーリエスペクトルと比較したものである．ここで，フーリエスペ
クトルはいずれも水平 2 成分のベクトル和をとりバンド幅
0.05Hz のパーセンウインドウ 70)を適用したものである（以下，本稿で示すフーリエスペクトルはいずれも同様）．この図から，相馬-G のスペクトルは 1.5Hz 付近では K-NET 相馬よりやや小さめであることがわかる．このような傾向が過去の地震にも表れているか検討するために，図-30～図-31 では 2003 年 5 月 26 日宮城県北部の地震（M7.1）および 2005 年 8 月 16 日宮城県東方沖の地震（M7.2）による
K-NET 相馬と相馬-G の記録の比較を行っているが，同様の傾向が表れている．今回の地震を含む 3 つの地震に対し事-GB）- 17 -

と小名浜事-G の記録の比較を行っているが，同様の傾向が

宮田さよ子さんには記録の処理などで大きく貢献していた表れている．
今回の地震を含む 3 つの地震に対して，K-NET

だいています．2010 年 6 月 13 日福島県沖の地震の記録を勿来に対する小名浜事-G のフーリエスペクトルの比を求紹介した章では，比較のため，防災科学技術研究所のめると図-36 のようになり，やはり 1.8Hz 付近では小名浜

K-NET の記録を利用しました．記して謝意を表します．

事-G の方がスペクトルが小さい傾向が表れている．この傾向が地震を問わず表れていることから，これは K-NET 勿来

参考文献と小名浜事-G におけるサイト特性の相違によるものと考

1）菅野高弘・三籐正明・及川研：兵庫県南部地震による港湾施設の被害考察（その 8）ケーソン式岸壁の被災に関えられる．

する模型振動実験，港湾技研資料，No.813，1995 年 9 月，

図-37 では相馬港の地表（相馬-G）と地中（相馬-GB）のフーリエスペクトルの比を求め，これを 2010 年 3 月 13

pp.207-252．

日の地震（M5.5）および 2010 年 3 月 14 日の地震（M6.7）

2）一井康二・井合進・森田年一：兵庫県南部地震におけると比較している．この図から，異なる地震に対してスペクケーソン式岸壁の挙動の有効応力解析，港湾技研報告，
第 36 巻，第 2 号，1997 年 6 月，pp.41-86．

トル比の形状はほぼ同様であり，地盤は線形の範囲で応答していると考えられる．図-38 は同様の検討を小名浜港を

3）野津厚・長尾毅：スペクトルインバージョンに基づく全対象に実施したものであるが，ここでもやはり地盤は線形

国の港湾等の強震観測地点におけるサイト増幅特性，港の範囲で応答していると考えられる．同様の検討を，地表湾空港技術研究所資料 No.1112, 2005年12月．

と地中で記録が得られている他の地点に対しても実施した

4）野津厚・菅野高弘：スペクトルインバージョンに基づくが，本地震に対して地盤が非線形挙動を示している地点は

道北の強震観測地点におけるサイト増幅特性，港湾空港

認められなかった．これらの観測結果は，今後，地盤の地技術研究所資料 No.1214, 2010年6月．
5）野津厚・盛川仁：表層地盤の多重非線形効果を考慮した

震応答計算プログラムの検証を行っていく上で有用であると考えられる．

経験的グリーン関数法，地震2，第55巻，2003年3月，
pp.361-374．

6. まとめ6）土田肇・山田逓一郎・倉田栄一・須藤克子：港湾地域

強震観測年報（1963･1964），港湾技研資料 No.55，1968
年 9 月．

本資料は，1962 年より実施されている港湾地域強震観測

7）土田の記録のうち，2010 年に得られた記録について報告するも肇・山田逓一郎・倉田栄一・須藤克子：港湾地域のである．得られた記録は地点毎に分類され，地震諸元，

強震観測年報（1965･1966），港湾技研資料 No.62，1968

記録番号，最大加速度等とともに付録 CD の強震観測表に

年 12 月．
8）土田

まとめられている．強震観測表の地震諸元は気象庁の地

肇・倉田栄一・須藤克子：港湾地域強震観測年報（1967），港湾技研資料 No.64，1969 年 3 月．

震・火山月報（カタログ編）に基づいている．強震観測表

9）土田にまとめられている記録の中で，最大加速度が 20Gal を越肇・倉田栄一・須藤克子：港湾地域強震観測年報（1968），港湾技研資料 No.98，1970 年 3 月．

える記録については加速度波形を示している．また，最大加速度が 50Gal を越える比較的大きな記録については，加10）土田肇・倉田栄一・須藤克子：港湾地域強震観測年

報（1969），港湾技研資料 No.100，1970 年 6 月．

速度波形に加え，速度波形，変位波形，フーリエスペクト

11）土田

ル，応答スペクトルも併せて示している．

肇・倉田栄一・須藤克子：港湾地域強震観測年

報（1970），港湾技研資料 No.116，1971 年 3 月．

（2011年8月15日受付）

12）倉田栄一・石坂徳三・土田肇：港湾地域強震観測年

報（1971），港湾技研資料 No.136，1972 年 3 月．

謝辞13）倉田栄一・石坂徳三・土田

港湾地域強震観測は，当所の他，国土交通省港湾局，東肇：港湾地域強震観測年

報（1972），港湾技研資料 No.160，1973 年 3 月．

北地方整備局，関東地方整備局，北陸地方整備局，中部地

14）倉田栄一・石坂徳三・土田

方整備局，近畿地方整備局，中国地方整備局，四国地方整

肇：港湾地域強震観測年

報（1973），港湾技研資料 No.181，1974 年 3 月．

備局，九州地方整備局，北海道開発局，内閣府沖縄総合事

15）倉田栄一・石坂徳三・土田

務局，東京都港湾局，静岡県港湾総室，大阪市港湾局，宮肇：港湾地域強震観測年

報（1974），港湾技研資料 No.202，1975 年 3 月．

崎県港湾課および国土技術政策総合研究所の各機関の協力の下に実施されています．
地震動研究チームの林公美さん，

16）倉田栄一・井合- 18 -

進・土田

肇：港湾地域強震観測年

1999 年 6 月．

報（1975），港湾技研資料 No.236，1976 年 3 月．
17）倉田栄一・井合進・土田36）佐藤幸博・一井康二・星野裕子・佐藤陽子・井合

肇：港湾地域強震観測年

報，補遺（1963~1975，上下動成分），港湾技研資料 No.250，

長尾1976 年 12 月．

料 No.942，1999 年 9 月．

進・

毅：港湾地域強震観測年報（1998），港湾技研資肇：港湾地域強震観測年

37）野津厚・深澤清尊・佐藤陽子・玉井伸昌・菅野高弘：報（1976･1977），港湾技研資料 No.287，1978 年 3 月．

港湾地域強震観測年報（1999&2000），港湾空港技術研究

18）倉田栄一・井合19）倉田栄一・井合進・土田進・横山淑子・土田所資料 No.1016，2002 年 3 月．

肇：港湾地域

38）深澤清尊・佐藤陽子・野津厚・菅野高弘：港湾地域強強震観測年報（1978･1979），港湾技研資料 No.338，1980
年 6 月．
20）倉田栄一・井合

震観測年報（2001），港湾空港技術研究所資料 No.1019，
2002 年 6 月．

進・横山淑子・野田節男：港湾地域

強震観測年報（1980），港湾技研資料 No.374，1981 年 639）野津厚・佐藤陽子・深澤清尊・佐藤泰子・菅野高弘：

月．

港湾地域強震観測年報（2002），港湾空港技術研究所資21）倉田栄一・野田節男：港湾地域強震観測年報（1981），
港湾技研資料 No.426，1982 年 6 月．

料 No.1054，2003 年 9 月．
40）野津厚・鈴木嘉秀・早田泰子・菅野高弘：港湾地域強22）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

震観測年報（2003），港湾空港技術研究所資料 No.1084，

報（1982），港湾技研資料 No.446，1983 年 6 月．

2004 年 9 月．

23）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

41）野津厚・菅野高弘：港湾地域強震観測年報（2004），港湾空港技術研究所資料 No.1109，2005 年 9 月．

報（1983），港湾技研資料 No.487，1984 年 6 月．

42）野津厚・菅野高弘：港湾地域強震観測年報（2005），港

24）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

湾空港技術研究所資料 No.1136，2006 年 9 月．

報（1984），港湾技研資料 No.519，1985 年 6 月．

43）野津厚・菅野高弘：港湾地域強震観測年報（2006），港

25）倉田栄一・福原哲夫・野田節男：港湾地域強震観測年

湾空港技術研究所資料 No.1164，2007 年 9 月．

報（1985），港湾技研資料 No.547，1986 年 6 月．
26）倉田栄一・井合44）野津厚・菅野高弘：港湾地域強震観測年報（2007），港進・野田節男：港湾地域強震観測年

湾空港技術研究所資料 No.1184，2008 年 9 月．

報（1986），港湾技研資料 No.588，1987 年 6 月．

45）野津厚・菅野高弘：港湾地域強震観測年報（2008），港

27）倉田栄一・野田節男・樋口豊志：港湾地域強震観測年

湾空港技術研究所資料 No.1207，2010 年 3 月．

報（1987），港湾技研資料 No.618，1988 年 6 月．
28）倉田栄一・井合進：港湾地域強震観測年報（1988），46）野津厚・菅野高弘：港湾地域強震観測年報（2009），港

港湾技研資料 No.649，1989 年 6 月．
29）倉田栄一・井合湾空港技術研究所資料 No.1223，2010 年 12 月．

進：港湾地域強震観測年報（1989）
，

47）土田

港湾技研資料 No.676，1990 年 6 月．
30）倉田栄一・井合その余震の港湾地域における強震記録，港湾技研資料No.80，1969 年 6 月．

進：港湾地域強震観測年報（1990），

港湾技研資料 No.705，1991 年 6 月．
31）倉田栄一・井合48）倉田栄一・井合進・土田肇：1978 年伊豆大島近海の地震の港 湾地域における強震記録， 港湾技研資料

進：港湾地域強震観測年報（1991），

No.317，1979 年 3 月．

港湾技研資料 No.727，1992 年 6 月．
32）松永康男・桜井博孝・森田年一・井合肇・倉田栄一・須藤克子：1968 年十勝沖地震と49）倉田栄一・井合進：港湾地域進・横山淑子・土田肇：1978 年宮

強震観測年報（1992&1993），港湾技研資料 No.776，1994

城県沖地震の港湾地域における強震記録，港湾技研資料

年 6 月．

No.319，1979 年 6 月．
進：50）倉田栄一・野田節男：昭和 57 年（1982 年）浦河沖地

港湾地域強震観測年報（1994），港湾技研資料 No.840，

震の港湾地域における強震記録，港湾技研資料 No.442，

1996 年 6 月．

1983 年 3 月．

33）宮田正史・佐藤幸博・一井康二・森田年一・井合34）佐藤幸博・一井康二・井合進・星野裕子・佐藤陽子・

51）倉田栄一・福原哲夫・野田節男：昭和 58 年（1983 年）

宮田正史・森田年一：港湾地域強震観測年報（1995&1996），

日本海中部地震の港湾地域における強震記録，港湾技研

港湾技研資料 No.909，1998 年 9 月．

資料 No.458，1983 年 9 月．

35）佐藤幸博・一井康二・井合進・星野裕子・佐藤陽子：52）倉田栄一・福原哲夫・野田節男：昭和 59 年（1984 年）
8月7日

港湾地域強震観測年報（1997），港湾技研資料 No.936，- 19 -

日向灘地震の港湾地域における強震記録，港湾技研資料 No.503，1984 年 12 月．
53）倉田栄一・野田節男・樋口豊志：昭和 62 年（1987 年）
12 月 17 日

千葉県東方沖地震の港湾地域における強震

記録，港湾技研資料 No.619，1988 年 6 月．
54）松永康男・桜井博孝・森田年一・井合進：1993 年釧路沖地震の港湾地域における強震記録，港湾技研資料No.777，1994 年 6 月．
55）松永康男・桜井博孝・森田年一・井合進：1993 年北海道南西沖地震の港湾地域における強震記録，港湾技研
資料 No.778，1994 年 6 月．
56）佐藤幸博・宮田正史・一井康二・森田年一・井合進：

1994 年北海道東方沖地震の港湾地域における強震記録，
港湾技研資料 No.853，1996 年 12 月．
57）佐藤幸博・一井康二・宮田正史・森田年一・井合進：

1994 年三陸はるか沖地震の港湾地域における強震記録，
港湾技研資料 No.892，1997 年 12 月．
58）佐藤幸博・一井康二・星野裕子・佐藤陽子・宮田正史・
森田年一・井合進：1995 年兵庫県南部地震の港湾地域における強震記録，港湾技研資料 No.907，1998 年 6 月．
59）土田肇・山田逓一郎・倉田栄一：港湾地域強震観測

地点資料（その 1），港湾技研資料 No.34，1967 年 11 月．
60）倉田栄一・土田肇・須藤克子：港湾地域強震観測地点資料（その 2），港湾技研資料 No.107，1970 年 12 月．
61）倉田栄一・石坂徳三：港湾地域強震観測地点資料（その 3），港湾技研資料 No.156，1973 年 3 月．
62）横山淑子・倉田栄一：港湾地域強震観測地点資料（その 4），港湾技研資料 No.298，1978 年 6 月．
63）横山淑子・倉田栄一：港湾地域強震観測地点資料（その 5），港湾技研資料 No.351，1980 年 9 月．
64）一井康二・佐藤幸博・佐藤陽子・星野裕子・井合

進：

港湾地域強震観測地点資料（その 6），港湾技研資料No.935，1999 年 6 月．
65）気象庁：地震・火山月報（防災編），2009 年．
66）気象庁：地震・火山月報（カタログ編），平成 21 年 12月，2010 年．
67）井合進・倉田栄一・土田肇：強震記録の数字化と補正，港湾技研資料 No.286，1978 年 3 月．
68）Susumu Iai and Eiichi Kurata: Integration of strong-motion
accelerograms， Proceedings of the 5th Japan Earthquake
Engineering Symposium，1978，pp.225-232.
69）Kinoshita, S. : Kyoshin Net (K-net), Seim. Res. Lett., Vol.69,
1998, pp.309-332.
70）大崎順彦：新・地震動のスペクトル解析入門，鹿島出版会，1994 年 5 月．

- 20 -

強震観測担当者（2010年１月～12月）
東北地方整備局
青森港湾事務所

玉山

具夫

斉藤

雅志

八戸港湾・空港整備事務所

滝沢

洋一

齊藤

聡洋

福田

良介

櫻庭

貞昭

菅原

将徳

佐藤

和敏

原茂

雅光

釜石港湾事務所
釜石港湾事務所

宮古港事務所

塩釜港湾・空港整備事務所

外久保

小名浜港湾事務所

根本

俊彦

箭子

英雄

佐藤

匡

秋田港湾事務所

齋藤

信之

菅原

将徳

酒田港湾事務所

須田

国男

百澤

康雄

佐藤

栄治

渡辺

義

小名浜港湾事務所

相馬港事務所

裕一

山田

裕之

瀬川

哲

石澤

武紘

北陸地方整備局
新潟港湾・空港整備事務所
伏木富山港湾事務所

宮崎賢治

金沢港湾・空港整備事務所

新湊出張所

村崎

内生蔵

清武

一樹

杉原

友和

敦賀港湾事務所

木原

寿明

波多野

笹森

秀一郎

笹森

秀一郎

石岡

勉

佐藤

達也

四戸

秀治

五十嵐

渉

石田

崇

中村

弘樹

吉田

明俊
對木

努

前田

千

澤田

隆志

雅

関東地方整備局
鹿島港湾・空港整備事務所
鹿島港湾・空港整備事務所

常陸那珂港事務所

千葉港湾事務所
京浜港湾事務所
中部地方整備局
清水港湾事務所
清水港湾事務所

下田港事務所日比智也

清水港湾事務所

御前崎港事務所

冨本

正

長瀬

和光

梅田

至人

三河港湾事務所
三河港湾事務所

衣浦港事務所名古屋港湾事務所

久保田靖子

名古屋港湾空港技術調査事務所

木全

啓介

吉村

藤謙

四日市港湾事務所

奥山

健司

船橋

香

和歌山港湾事務所

森田

正樹

竹村

知子

神戸港湾事務所

渡慶次

仲濱

弘平

近畿地方整備局
賀則

中国地方整備局
境港湾・空港整備事務所

西丸

剛史

広島港湾・空港整備事務所

石橋

伸司

- 21-

四国地方整備局
松山港湾・空港整備事務所

田村剛

小松島港湾・空港整備事務所

斉藤祐一

高知港湾・空港整備事務所

真砂

澄男

的野

一郎

山本

弘幸

別府港湾・空港整備事務所

小野章

宮崎港湾・空港整備事務所

森満

賢治

志布志港湾事務所

野田巧

鹿児島港湾・空港整備事務所

井上

光輝

泉保

佑介

松尾

博文

山本

龍三

冨安

真一郎

竹口

はや人

貞島

一雄

本山

賢司

上村

剛志

九州地方整備局
博多港湾・空港整備事務所

荻定治

北海道開発局
釧路開発建設部釧路港湾事務所

山内弘明

釧路開発建設部根室港湾事務所菊地聡一

帯広開発建設部十勝港湾事業所

八栁

敦

鈴木

孝信

室蘭開発建設部

浦河港湾事務所

伊東

公人

山岡

広幸

室蘭開発建設部

苫小牧港湾事務所

渡邊

毅人

鈴木

貴秀

室蘭開発建設部室蘭港湾事務所

大沢佳宏

小樽開発建設部小樽港湾事務所

杉山

盛行

松本

隆一

三浦

豪

函館開発建設部函館港湾事務所

長嶺

鎌弘

明石

卓弥

函館開発建設部

江差港湾事務所

川西

健孔

藤由

幸大

留萌開発建設部留萌港湾事務所

佐々木

網走開発建設部

網走港湾事務所

香川智章

網走開発建設部

紋別港湾事務所

鈴木

慶律

智之

沖

享尚

沖縄総合事務局
那覇港湾・空港整備事務所
那覇港湾・空港整備事務所

中城湾港出張所

佐々木

暢彦

伊良波

憲

中岡

泰典

城間

由樹

平良港湾事務所

国場

幸恒

石垣港湾事務所

田港

朝之東京都港湾局下拂

章一森

大阪市港湾局

富田

太朗

鈴木

健司

黒田

剛

静岡県田子の浦港管理事務所

佐野光政

宮崎県土木部北部港湾事務所

橋口

公進

佐藤

博昭

黒木

成一

児玉

勝美

都道府県

-22-

洋

港湾空港技術研究所資料

No.1243２０１１．１２

編集兼発行人

独立行政法人港湾空港技術研究所

発

行

所

独立行政法人港湾空港技術研究所
横 須 賀 市 長 瀬 ３ 丁 目 １ 番 １ 号
TEL. 046(844)5040 URL. http://www.pari.go.jp/

印

刷

所

株 式 会 社 シ ー ケ ン

C （2011）by PARI
Copyright ○All rights reserved. No part of this book must be reproduced by any means without the written
permission of the President of PARI
この資料は、港湾空港技術研究所理事長の承認を得て刊行したものである。したがって、本報告書の全部または一部の転載、複写は港湾空港技術研究所理事長の文書による承認を得ずしてこれを行ってはならない。