SINGLE KERNELMOISTURE TESTER

CTR-500F

seriesStart / Stop

Fast moisture distribution
analysis at a timeof receiving grains.

PaddyWheat

Buckwheat

Outcome of poor drying grains can quickly be found on LCD screen
Instantly determine moisture distribution. Green, Yellow, and Red color result provides easy-to-read moisture distribution
at a glance. Variation of moisture distribution on grains at a glance on LCD screen. Drying time is varied by moisture
content and variation of grains. Checking variation of moisture distribution of grains will prevent over drying accident.
GreenSmallnumbers

Yellow

Red

Moisture content and variation of grains

Largenumbers

Fast Measurement for paddy

Single Kernel Measurement

You can greatly reduce the working hours by achieving double
speed of normal mode with “High Speed Mode for paddy”.

One kernel at a time measurement provides moreaccurate reading than handheld moisture meter.

Features
Intuitive Color
Touch screen

Development and modiﬁed
Calibration Curve

Intuition tells you that getting on
Simple and easy to use with large
LCD screen.

Custom calibration could be created with
CTR-500F series application.
Several kinds of custom calibration could
be registered onto CTR-500F series.
Application is available on
https://www.shizuoka-seiki.co.jp/english/

Data can be saved onto
SD memory card（option）

Measures wide variety of grains（Compatible with long grain species）

Data Storage and Capacity:
Data can be saved onto SD memory card.
For instance,
Memory storage for up to 2.6 million
measurements results/1GB.
Date could be exported to PC.

CTR-500F series is capableof measuring Chinese long kernel,
Indonesian long kernel: paddy, brown
rice, and white rice.

Wide moisturemeasurement range

Internal PrinterHistogram, single kernel result, andmany types of result could be printedout.

CTR-500F series provides measurementof paddy and wheat with up to 40%
moisture.
High moisture content of grains canalso be measured.

How to use（Easy to Operate）
1

Place Kernel
sample2Starting
measurement

3

Measurement on
LCD screen

4

Results display

5

Remaining kernels
handling process

Alterable

Place Kernel sample intomeasurement basket.

Press Start / Stop button,then measurement start.

Counting quantity of kernels.

Results of both averagemoisture and histogram aredisplaced on LCD screen.

Remaining kernels andcrushed kernels are
separated into diﬀerent trays.

Main speciﬁcations
Model
Power source
Display character [ Initial setting]Selectable language
Print character [ Initial setting]Selectable language
Method：Electrode：
Display type：
Printing method：
Moisture content fine adjustment
Power consumption:
Dimension(W/H/D)
Weight:Kernel types and measuring range（The ambient temperature is 20℃）

CTR-500FE
CTR‐500FH
AC220V
AC220V
EnglishHangeulTraditional ChineseEnglishSimplified Chinese
Traditional Chinese
JapaneseSimplified Chinese
HangeulJapanese
EnglishHangeulTraditional Chinese
N/A
Simplified Chinese
Japanese
Direct current resistance method
2 roller, Rotation ratio 2:1
Color liquid crystal touch panel
Thermal type-2.0% ～＋2.0%
120VA
246×310×293 mm (Not including protrusions).
11.2kgBrown rice(produce of Japan)
11.1% - approximately 20%
White rice(produce of Japan)
11.4% - approximately 20%
Wheat(produce of Japan)
10.2% - approximately 40%
Barley(produce of Japan)
9.0% - approximately 36%
Rye(produce of Japan)
11.4% - approximately 30%
Buckwheat(produce of Japan)
11.1% - approximately 30%
Long brown rice (Indonesian long kernel, Chinese long kernel)
10.2% - approximately 20%
Long white rice (Indonesian long kernel, Chinese long kernel)
10.6% - approximately 20%
Standard
10.0% - approximately 45%
In drying paddy(produce of Japan)
11.0% - approximately 40%
Paddy(produce of Japan)
10.0% - approximately 40%
Long paddy(Indonesian long kernel, Chinese long kernel)
9.2% - approximately 40%
CTR-500FS
AC115V
Traditional Chinese
Simplified ChineseEnglishJapaneseHangeulTraditional Chinese
Simplified ChineseEnglishJapanese＊Some types of grain may have diﬃculty analyzing. ＊Measuring range may diﬀer by ambient temperature◇SPECIFICATIONS ARE SUBJECT TO CHANGE WITHOUT NOTICE.

1300 Moroi, Fukuroi ‒Shi , Shizuoka-ken 437-1121 JAPAN
Phone：+ 81-538-23-3990 Fax：+ 81-538-23-3192
E-mail：international@shizuokaseiki.com
URL：https://www.shizuoka-seiki.co.jp/english/