PAPERLESS RECORDER

GR200 SERIES

Larger Display=Enhanced Visibility

Portable unit available

144

Various displays via one-touch key operation

160

185

Easy operation
Large indication on a 5.7" display

RecorderView your recording data easily

Advantages of using the paperless recorder■ Your PC can easily manage recorded data

■ Running costs reduced

It is easy to store recording data and create
reports.
Recording data can be managed using EXCEL.

■ No malfunctions due to operational abrasion

Consumables such as chart paper, inks
are not necessary, more streamlined
operation, save time and money.

■ Record more for longer

Malfunctions due to mechanical abrasion are
avoidable as motors, gears, etc. which are requiredfor chart paper recorders are no longer necessary.

Using a 1GB CF card, data can be recordedfor 6 years. (Using a 30sec recording cyclein the case of 9-point recording)■ Multi-point recording
9-point, 18-point available

Shinko paperless recorder features …
● Easy operation● Large 5.7" display● Various displays via one-touch key operation

Trend recording (horizontal)

Trend recording (vertical)

Bar graph display■Other displays available:● Event summary display● Ethernet log display● Parameter display● Tag display● Historical trend display

Digital (totalizing data) display

Analog meter display

Name of sections① Status indication
Indicates the display name, calendar, alarm information, recording
status, compact ﬂash writing status, compact ﬂash loading status, etc.
② Time indication
Indicates measured time and its scale.

③ Trend indication
Indicates measurement results using a waveform.

④ Channel indication
Indicates measurement values numerically.

⑤ Function keyboard
Recording Start / Stop, indication switching, setting, dataindication / alteration, etc. can be conducted.

■ Optional specifications
User specified• DI / D0 (10 )Communication
function [C5]: Relay output 10-points + DI input 5-points• DI / D0 (18 ), C5 : Open collector output 18-points +DI

Alarm output[DI/D0(10)][DI/D0(18),C5]

input 5-points + RS485 communication
Relay output (10-points ) : 1a contact, contact capacity• D01: 3A 150V AC ( resistive load ) or 3A 30V DC ( resistive load ): 3A 240V AC ( resistive load ) or 3A 30V DC ( resistive load )• D02 to D10
No-voltage contact input (5-points ) :Communication interface
ProtocolCommunication Speed
Transmission speed
Transmission method: EIA RS-485: MODBUS (RTU): 9600, 19200bps: 10Mbps: Baseband

Max. network length or Max. node interval : 500m (4-level cascade)Max. segment length: 100m (Between node-HUB)
Connecting cable: UTP (Unshielded, twisted pair cable)

Ethernet[E - net]

ON pulse width : 200msec or more
OFF pulse width : 200msec or more
Open collector output (18-points ) : 0.1A 30V DC

22-26AWG: TCP / IP

Protocol■ Optional specifications (Scale: mm)

171.5

161
21.6160

185

160

Foot stand24.6 (When panel packing is used)

21.6

Foot3415144

37Handle【Portable】

Mounting bracket

144【Panel mounting】

185■ Input terminal arrangement【9 - points input】■ Alarm output /DI input terminals【18-points input】■ DIO terminals (D-Sub)( Input 9-points only )■ Power terminals■ Panel cutout (Scale: mm)【Single unit mounting】【Panel mounting】 【Portable】
LN

137 ＋1.50100〜240V AC
50/60Hz

137 ＋1.50100〜240V AC
50/60HzCOMMUUNICATIONDIO TERMINAL【Lateral close mounting】

137 ＋1.50(160×n- 22) ＋20● To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
● This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify！

SAFETY
PRECAUTIONS

correct usage after consulting purpose of use with our agency or main oﬃce.
(Never use this instrument for medical purposes with which human lives are involved.)● External protection devices such as protection equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Also proper
periodic maintenance is required.
● This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in this manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as acomponent in, or as being utilized in themanufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that this
instrument is not illegally exported.

・This catalog is as of July 2008 and its contents are subject to change without notice. ・If you have any inquiries, please consult us or our agency.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Reg. Oﬃce : 2-5-1, Senbahigashi, Minoo, osaka, 562-0035, JapanTel: 81-72-727-6100

Fax: 81-72-727-7006

URL: http://www.shinko-technos.co.jp

E-mail: overseas@shinko-technos.co.jp
78-201-GR2 (E)