DIGITAL BORE GAUGE

Instruction Manual

Model No.：WCDI-50D/WCDI-100D/WCDI-160D <combined>

Thank you for purchasing the Digital Bore Gauge.
Please read this manual thoroughly before use for proper operation.

■PART IDENTIFICATIONGauge

LCD Display(Digital Indicator)[ABS/SET]
ButtonABS

Shaft

Mounting
Nut
Extension
RodLength below Handle:150mm【Accessories】
0.5mm 1mm 2mm 3mm
Washers

Sub Rod 55mm

（WCDI-160D only）

Probe Tip
Measurement
HeadGuide■ INSERTING・REPLACING BATTERYNOTICE●Extension Rods

WCDI-50D････35・40・45・50mm（４pieces）
WCDI-100D･･･50・55・60・65・70・75・80・85・90・95・100mm（11 pieces）
WCDI-160D･･･50・55・60・65・70・75・80・85・90・95・100・105mm（12 pieces）●Sub Rod･････55mm×１x（WCDI-160D only）●Washer･････0.5・1・2・3mm (1 each size)●Battery･･････CR2032Lithium button cell, 1x (for test)

Battery（CR2032）

Battery
Case

Use only CR2032 typelithium button cell.

■ AUTO OFF DELAY①Press and hold the [ZERO/ON] Button to turnpower OFF.
②Press and hold [ABS/SET ] and [ZERO/ON]
Buttons at the same time.
③When「-----」 is displayed, ❶release the[ZERO/ON] Button, ❷then release the[ABS/SET ] Button in that sequence.
④The Auto Off Delay time will be displayed. Press[ABS/SET] to change.
※Each press of the [ABS/SET ] Button will change
the delay by 1/2 hr. (30 min.) from 0 to 6 hr.

⑤Press [ZERO/ON] when done and display will
return to the measurement screen.

※Once set, the delay time is saved even when poweris OFF.

■ ASSEMBLY①Loosen the Clamp Screw and insert the DigitalIndicator into the gauge body. Tighten the Clamp
Screw to secure.
※After attaching, press the [ZERO/ON] Button on theDigital Indicator to turn on power. Confirm that thedisplayed reading changes by gently squeezing themeasurement head. If necessary, adjust the depththat the Digital Indicator is inserted into the gaugebody.

ABS/SETMIN

ZERO/ONCAL②Select the combination of Sub Rod (WCD-160
Screwonly) Extension Rod and Washers for thedimension you want to measure.
Remove the Rod Adj. Nut put on the desired Sub Rod, Rod, and Washers,
and secure with the Extension Rod Mounting Nut.
Clamp

※Please select combination with fewest number of Washers.

【Examples】

Ex. Rod Nut

Ex. Rod 60mm

To
Measure:WasherWasher
Ex. Rod＋ 0.5mm
60mm ＋2mm

ToMeasure:118 mm

Ex. Rod ＋ Washer ＋3mm60mm※Measure absolute dimension of workpiece” or “” symbols are displayed on the LCD, press the [MIN] Button to turn off.

②Adjust the “SET” value of the Bore Gauge to match the ID of the Ring Gauge Standard. Press and hold the [ABS/SET] Button until“SET” blinks on the display.
③Continue holding the [ABS/SET] Button and the display character to the left will blink. Use a short press of the [ABS/SET] Button tochange the value, and a long press to change the character being modified. Repeat this process to enter the dimension of the Ring
Gauge Standard on the LCD display. (Figure 4)④When the value is entered the flashing position will return to “SET”. A shortpress of the [ABS/SET] Button to turn off the flashing.

⑤Insert the Measurement Head into the Ring Gauge Standard and press the[MIN] Button to put the Indicator into minimum-mode. Slowly pivot thegauge (Figure 3) to display the minimum value.
⑥Remove the gauge from the Ring Gauge and press the [CAL] Button until“OK” is displayed.
Preparation for measurement is now complete.

Sub Rod
55mmABSMIN



OO

ABS/SETZERO/ON

Set to match

CAL

Ring GaugeStandard（Fig. 4）⑦Put the gauge into the workpiece to be measured, and pivot the gauge
slowly. (Figure 3) The LCD will display the measured value. (Figure 5)※Press the [ON/ZERO] Button momentarily to reset display in preparation
to measure the next workpiece.

COMPARATIVE MEASUREMENTS ※Difference between workpiece and reference①Turn on power. If “” or “” symbols are displayed on the LCD,
press the [MIN] Button to turn off. If “ABS” is displayed press the[ABS/SET] Button to turn off.

ABSOO

ABS/SETMIN

ZERO/ON※Press the [ON/ZERO] Button momentarily to reset display in preparation
to measure the next workpiece.

■ TROUBLESHOOTINGERRORCORRECTIVE ACTION●LCD does not display●Displayed value is unstable●Display is not clear●Cycle power by removing andreplacing battery.
●Replace battery with new one.

Workpiece I.D. value

CAL

Workpiece（Fig. 5）②Insert the Measurement Head into the Ring Gauge Standard and press[MIN] Button to put the Indicator into minimum-mode. Slowly pivot thegauge (Figure 3) to display the minimum value.
③Remove the gauge from the Ring Gauge and press the [CAL] Button until“OK” is displayed.
Preparation for measurement is now complete.
④Put the gauge into the workpiece to be measured, and pivot the gaugeslowly. (Figure 3) The LCD will display the difference between themeasured workpiece and the calibration standard. (Figure 6)

To
Measure:62.5 mm（Fig. 3）①Before use, please make sure the Measurement Head components are not loose and tighten if necessary.
②Clean the Measurement Head and the ID of a Ring Gauge Standard using parts cleaner or cleaning solution.
③Press the [ZERO/ON] Button to turn on power. ※ A long press to the [ZERO/ON] Button will turn power off.

ABSOLUTE MODE (ABS)●This is a precision instrument, handle with care.
●Do not disassemble or modify.
●Keep instrument away from water and oil.
●Use only with supplied Extension Rods and Washer.
●Keep instrument away from direct sunlight and high temperatures such as ina car, or near a stove or heat source.
●After use, remove any dirt or cutting chips and apply rust preventative oil toExtension Rod and Washer. When not in use store in supplied storage case.
●This product is for measuring inside diameter; use only as directed. Improper
use may cause accident or injury.
●Avoid using in high electric fields, such as near fluorescent lights or switching
power supplies. Electrical interference may cause incorrect readings.

6 0 mm（Fig. 2）

①Turn on power. If the “INSERTPull out the case fromthe two side edges toremove. Insert batterywith (+) side facing
down.

（Fig.1）■ OPERATIONACCESSORIESNOTICE

Extension RodｂOO

Handle570ａZERO/ON※1：Not Used※2："0.01" indicates theresolution of the display's
digital bar graph. The bargraph display will changewith each 0.01mm change
in the measured value.

Cap

WCDI -160D
5 0 - 16 00.002
1863
1. 6CAL[ZERO/ON]
Button[CAL] Button

WCDI -100D
5 0 - 10 00.002
1863
1. 60 〜 6.0475

ABS/SET

ZERO/ON
CAL

Model No.
WCDI -50D
Measuring Range （ｍｍ）
3 5- 5 00.002（ｍｍ）
Resolution※＊ Wide range accuracy
15（μｍ）※＊ Adjacent error
5（μｍ）※＊ Repeatability
3（μｍ）
1. 2
Single Stroke Range（ｍｍ）（ｈｒ）
Auto Off Delay（ｇ）
Weight450

ABSABS/SETMIN[MIN] Button
ClampScrew※2＊：Including supplied Digital Gauge※Not including quantization error (±1 count)

MIN

※1

OO

■ SPECIFICATIONS■ MEASUREMENT
For accurate measurements, follow the procedure described below.
①In cross section perpendicular to the axis of the bore, as shown in Figure 1, position the gauge along diameter ⓐ. This is themaximum distance, and at this point the Indicator will read the maximum value. The Measurement Head will automatically move todiameter ⓐ when inserted.
②In cross section parallel to the axis of the bore through diameter ⓐ, position the indicator to minimize the distance ⓑ. (Figure 2)
In this position the Indicator will read the minimum value. This position must be manually determined by pivoting the Gauge asshown in Figure 3 while watching the Indicator to find the minimum value.

OO
ABS/SETMIN※If problem persists, or if you have anyquestions, please contact distributor.
※Please note, manufacturer is unable torespond to inquires or provide service
directly. Please contact distributor.

ZERO/ONCAL

Diff. between Workpiece and Reference

Workpiece（Fig.6）

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

デジタルシリンダゲージ‫ٯ‬ඏ‑ད࿉‫۽‬ᘹ⬀ nvq p nrqq p nrwq a⬂Ԁᘈ⬄

この度はデジタルシリンダゲージをお買上げ頂きありがとうございます。
ご使用前に、本書をよくお読み頂き正しくお使いくださいますようお願い申し上げます。

■各部の名称■仕 様指示器

LCD表示部

CALボタン

取付ねじ※1：このボタンは使用しません。
※2：「0.01」表示は液晶表示部外周のバーグラフの単位です。
測定値に0.01mmの変動があった場合それに応じて、バーグラフも1本増減します。

キャップ

にぎり

【付属品】
0.5mm 1mm

換えロッド
取付ねじ

ワッシャ

にぎり下長さ150mm

外筒2mm 3mm

換えロッド

サブロッド55mm（WCDI-160Dのみ）

案内板■電池のセット・交換

電池蓋両側をツマミながら引き出し、(＋)
側が下になるよう電池をセットして、電池蓋を戻してください。

WCDI-50D････35・40・45・50mm（計４本）
WCDI-100D･･･50・55・60・65・70・75・80・85・90・95・100mm（計11本）
WCDI-160D･･･50・55・60・65・70・75・80・85・90・95・100・105mm（計12本）●サブロッド･････55mm×１本（WCDI-160Dのみ）●ワッシャ･･･････0.5・1・2・3mm（各1枚）●電 池･････････CR2032（コイン型リチウム電池：テスト用）×１個電池（CR2032）

電池蓋

電池は必ずCR 2032（コイン型リチウム電池）をお使いください。

■オートオフの時間設定①ZERO/ONボタンを長押しし、電源をOFFにします。
②ABS/SETボタンとZERO/ONボタンを同時に押します。
③LCD表示部に「-----」が表示されたら、❶
ZERO/ONボタン→❷ABS/SETボタンの順に放します。
④LCD表示部に設定時間が表示されるので、ABS/SETボタンを押し、希望の時間を表示させます。
※ABS/SETボタンを押すたびに「0.5時間（30分）」ずつオートオフ時間が切り替わります。
（0〜6.0時間）⑤ZERO/ONボタンを押し、画面が測定画面に切り替わったら設定完了です。
※電源をOFFにしても、設定時間は記憶されています。

● 換えロッド・ワッシャなどは付属のものを使用し、他との併用は避けてください。

● 直射日光の当たる場所や炎天下の車の中、火のそばなどに放置しないでく● 使用後は柔らかい布などでゴミや切粉などを取り除き、換えロッド・ワッシャなどは防錆処理をして、収納ケースの元の位置に保管してください。
● 本製品は内径測定用機器です。用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。
● インバータ式蛍光灯などの電界が発生する場所でのご使用は避けてください。LCD表示部の数値が誤作動を起こす原因となります。

■本体の組み付け①取付ねじをゆるめ、指示器を本体に差し込みます。取付ねじを締めて固定してください。

ABS/SETMIN※固定後、ZERO/ONボタンを押して電源を入れ、測定子を押してLCD表示部の数値が変わることを確認してください。上手く動作しない場合は、指示器の差込深さを再度調整してください。

②測定する寸法に合わせて、サブロッド（WCDI160Dのみ）・ 換えロッド・ワッシャを選択します。換えロッド取付ねじを外し、サブロッド・換えロッド・ワッシャを本体に取り付け、換えロッド取付ねじで固定してください。

取付ねじ※ワッシャの数量がなるべく少なくなるように選択してください。
【例】
６０ｍｍ
測定用

換えロッド
取付ねじ

換えロッド
60mm ＋

換えロッド 60mm

ワッシャ
2mm＋

ワッシャ
0.5mm

②測定子を含めた測定部と、基準リングゲージの内径をパーツクリーナーなどの洗浄液で洗浄します。
③ZERO/ONボタンを押して、電源をONにしてください。※電源をOFFにする時は、ZERO/ONボタンを長押ししてください。

①電源をONにします。LCD表示部に『● 分解・改造はしないでください。

【例】
１１８ｍｍ
測定用

■使用方法

ワッシャ
3mm＋サブロッド
55mm』のマークが表示されている場合は、MINボタンを押してマークを消してください。

ZERO/ONCAL③長押しを続けると、点滅位置が変わって行きます。変更したい数値の位置に点滅が移動して来たら、一度ABS/SETボタンを放し、今度は短く押してください。数値が変わります。
これを繰り返し、マスタとなるリングゲージに表示されている実寸法と、LCD表示部の数値を一致させてください。（図４）④数値が一致したら点滅位置を『SET』まで戻し、ABS/SETボタンを短く押して点滅を解除してください。
⑤測定部をマスタとなるリングゲージに入れ、MINボタンを押して最小値測定モードにします。（図３）のようにシリンダゲージをゆっくりと振り、最小値を表示させてください。
⑥測定部をマスタとなるリングゲージから抜き、CALボタンを長押ししてください。『OK』が表示されます。
これで測定の準備が完了しました。

ABSMIN

OO

ABS/SET

ZERO/ON

数値を合わせるCAL

マスタとなる
リングゲージ（図４）⑦ワークに測定子を入れ、（図３）のようにシリンダゲージをゆっくりと振って、測定してください。LCD表示部に、測定値が表示されます。
（図５）※次のワークを測定する際は、ZERO/ONボタンを一度押してください。

比較測定※ワークのサイズをマスタとなるリングゲージとの差で表示します。

①電源をONにします。LCD表示部に『ABS』、『』、『』のマークが表示されている場合は、ABS/SETボタンとMINボタンを押してマークを消してください。

ABS

OO

ABS/SETMIN

ZERO/ON
CAL

ワークの内径を表示②測定部をマスタとなるリングゲージに入れ、MINボタンを押して最小値測定モードにします。
（図３）のようにシリンダゲージをゆっくりと振り、最小値を表示させてください。

④ワークに測定子を入れ、（図３）のようにシリンダゲージをゆっくりと振って、測定してください。LCD表示部に、マスタとなるリングゲージとの差が表示されます。（図６）※次のワークを測定する際は、ZERO/ONボタンを一度押してください。

■トラブルと対応
トラブルの内容● 表示が安定しない● 表示が不明確である

対応策● 電池を入れ直してください。
● 新しい電池と交換してください。

ワーク

（図５）

③測定部をマスタとなるリングゲージから抜き、CALボタンを長押ししてください。『OK』が表示されます。
これで測定の準備が完了しました。

● ＬＣＤ表示部に何も表示されない

換えロッド60mm ＋』、『②マスタとなるリングゲージの数値に、シリンダゲージの数値を合わせます。ABS/SETボタンを長押ししてください。
『SET』マークが点滅します。

● 精密機器ですので、落としたりぶつけたりしないでください。

【例】
６２.５ｍｍ
測定用（図３）

絶対測定（ＡＢＳ） ※ワークのサイズを絶対値で表示します。

ご注意

差し込むご注意●換えロッド

ださい。

測定子

（図２）

①ご使用の前に、測定部と外筒の間がゆるんでいないか確認し、ゆるんでいたら締め付けてください。

● 水や油のかかるところで使用しないでください。

換えロッド

測定部

570

付属品

（図１）

OO

MINボタンｂZERO/ON

ZERO/ON
ボタン

ZERO/ON
CAL

ａCAL

OO

ABS/SETMINW C D I -16 0 D
5 0 -16 00 . 0 02
1863
1. 6ABS/SET※1※2※ 量子化誤差（±１カウント）含まず

W C D I -10 0 D
5 0 -10 00 . 0 02
18631. 60 〜 6.0475②直径䒏を通る円筒の軸に平行な断面において、直径书は最小値となり、指示器は最小値を表示します。（図２）
この直径书は、手動で求めます。シリンダゲージを（図３）のように振り、指示器が最小値を示す点を探してください。

ABSABS品
番
WCDI-50D
35 - 5 0測 定 範 囲（ｍｍ）
0 . 0 02
目量（ｍｍ）※＊
15
広 範 囲 精 度（μｍ）※＊
隣 接 誤 差（μｍ）
5※＊3
繰 返 し 精 密 度（μｍ）
1. 2
測定子の有効ストローク（ｍｍ）
オートオフ設定（時間）
質量 （ｇ）
45 0①円筒の軸に垂直な断面において、直径䒏は最大値となり、指示器は最大値を表示します。（図１）
この直径䒏は、可動する案内板によって自動的に求められます。

MIN

ABS/SET
ボタン（デジタルインジケータ）＊：付属のデジタルインジケータ（指示器）を含めた値です。

■測定方法について

正確な数値を測定するために、測定は下記の手順で行ってください。

OO
ABS/SETMIN

ZERO/ONCAL※症状が改善されない場合や不明な点がございまし
たらお買い上げの販売店、または当社までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

マスタとなるリング
ゲージとの差を表示

ワーク

（図６）

〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp2401

I384-K2