ADDENDA

Programmable controller

Model:

PC-935 T1633

The PC-935-x/M T1633, Programmable Controller, differs from the standard PC-935 in the following points.
For more specification details, please refer to the Catalog “PC-900 Series”.
Program performance
Number of patterns
16 (Linkable), Indication: 0 to 9, A to F
Number of steps
256 (16 steps/Pattern), Indication: 0 to 9, A to F
Number of repetitions
0 to 9999 times (No repeat action when set to 0)
Program time range
0 to 99 hours 59 minutes/Step, or 0 to 99 minutes 59 seconds/Step
Time setting accuracy
Within 0.1% of setting time
(0 to 100) ( ), With decimal point: (0.0 to 100.0) ( )(No Wait action when set to 0 or 0.0)
Wait value
DC current, voltage input: (0 to 1000) (The placement of the decimal point follows theselection.)
Spec: External selection/operation: The following functions are operable with Contact (or open collector) CLOSED.
Pattern numbers 0 to 15 are selectable by connecting external terminals.
TerminalNo.
0 (*)
Pattern number
external selection(13) b0(14) b1(15) b2(24) b3

STOPRUNADV (ADVANCE)
HOLD

12

3

OOOO45

Pattern number
6 7 8 9 10OO

O

O
OO
O
OOO

O

11O

O
O

O

O

1213

1415O
OO
OO
OO

OO

O

O

O

O: Selects a pattern number by connecting b0/b1/b2/ b3 to COM terminal.
(*) When you want to use the maximum number of External selections, select “Pattern 0”
by keypad operation.
Stops the running program control.
Starts the program RUN.
Interrupts performing step, and moves to the next step.
Not available

Terminal arrangement
GND: Ground terminal
POWER SUPPLY: 100 to 240V AC or 24VAC/DC
C1: OUT1(Control output 1) or Heating OUT1(D option)
C2/A2/LA: Cooling OUT2 (D option), Alarm 2 output(A2 option), Pattern end output A2 (A2 option),
Loop break alarm (LA option)
A1: Alarm 1 output or Pattern end output (A1)
RS-232C(RS-485): Communication RS-232C (C option),
RS-485(C5 option)
PATTERN SELECT: Pattern No. external selection
TC: Thermocouple input
RTD: RTD input
DC mA : DC current input
DC V : DC voltage input
TRANSMIT OUTPUT: Transmission output (TA, TV option)
EXT CON INPUT: External operation input
A4: Alarm 4 output or Pattern end output (A4)A3
: Alarm 3 output or Pattern end output (A3)
T.SIGNAL OUTPUT: Time signal output (TS option)

SHINKO TECHNOS CO., LTD.

URL: http://www.shinko-technos.co.jp/e/

TEL: +81-72-727-6100
FAX: +81-72-727-7006
E-mail: overseas@shinko-technos.co.jp

20100603

ADDENDA

Programmable controller

Model:PC-935 T1634

The PC-935-x/M T1634, Programmable Controller, differs from the standard PC-935 in the following points.
For more specification details, please refer to the Catalog “PC-900 Series”.
Spec: External selection/operation: The following functions are operable with Contact (or open collector) CLOSED.
Pattern numbers 0 to 9 are selectable by connecting external terminals.
TerminalNo.
Pattern number
external selection

0 (*)(13) b0(14) b1(15) b21

Pattern number
2 3 4 5 6O
OOOOO(24) b3OO
O

78O
O
O

9OO

O

O: Selects a pattern number by connecting b0/b1/b2/ b3 to COM terminal.
(*) When you want to use the maximum number of External selections, select “Pattern 0”
by keypad operation.
STOPRUNADV (ADVANCE)
HOLD

Stops the running program control.
Starts the program RUN.
Interrupts performing step, and moves to the next step.
Not available

Terminal arrangement
GND: Ground terminal
POWER SUPPLY: 100 to 240V AC or 24VAC/DC
C1: OUT1(Control output 1) or Heating OUT1(D option)
C2/A2/LA: Cooling OUT2 (D option), Alarm 2 output(A2 option) or Pattern end output 2 (A2 option),
Loop break alarm (LA option)
A1: Alarm 1 output or Pattern end output (A1)
RS-232C(RS-485): Communication RS-232C (C option),
RS-485 (C5 option)PATTERN SELECT: Pattern No. external selection
TC: Thermocouple input
RTD: RTD input
DC mA : DC current input
DC V : DC voltage input
TRANSMIT OUTPUT : Transmission output (TA, TV option)
EXT CON INPUT: External operation input
A4: Alarm 4 output or Pattern end output (A4)A3
: Alarm 3 output or Pattern end output (A3)
T.SIGNAL OUTPUT: Time signal output (TS option)

SHINKO TECHNOS CO., LTD.

URL: http://www.shinko-technos.co.jp/e/

TEL: +81-72-727-6100
FAX: +81-72-727-7006
E-mail: overseas@shinko-technos.co.jp

20100603