2-D Luminance Uniformity Analyzer

ColorLive CL-300
For luminance and chromaticity uniformity measurement of variously sized flat
display such as Back light, Front light, LCD, Flat light, OLED and Digital signage.

Features and Sales points
Wide measurement range
Luminance and Chromaticity distribution image available
Compact and lightweight
Size 100×45×45mm Camera body 216g
Multipoint measurable 1936×1216 points max.
Moire reduction function when measuring LCD
Suitable for both development and mass production lines※Camera stand is soldseparately.

Automatic judgment function, and external control by customized software
behave effectively in your mass production line!
■Data saving■Easy installation

Easy installation by USB connection of camera and memory key. After copy Measured data can be saved by the following file format.
Graphical content : JPEG PNG BMP Measured value : CSV Original : CLIof software into your PC, it can be started without USB memory key.
Re-measurement from the original file(CLI) is possible.
■Automatic judgment of acceptance/rejection
Measurement results can be saved automatically in sequential form of last 4
Judgement of acceptance/rejection can be made by setting the conditions digits of 0001~9999.
of optical specifications. If measured results are out of specifications, the ■Custom modification of software
values are displayed in red color.
Customized software is available in order to externally control the timing of such
After measurement, “Pass” or “Fail” can be displayed in large lettersmeasurements and data savings using digital input/output boards etc.
automatically.

■Wide-angle lensDistortion correction and peripheral illumination
correction let you measure in full-screen.

■Chromaticity display switching (x,y / u’,v’)■Keylock
In order to prevent malfunctioning or incorrect
setting, the system can be administered by password.

■Multiple languages
English, Chinese and Japanese■Measuring specifications
Name and No.

ColorLive

CL-300ContentsLens

Focal length 12.5mm

Camera2.35million pixel (1936×1216 pixel) 1/1.2 inch CMOS sensor

Resolution of camera

RGB 4096 gradations each（12bit each）

Luminance measurement
range（*1）

Type L

0.1 〜 30,000 cd/㎡

Type H5

Luminance linearity（*1）±3％Chromaticity accuracy（*1）±0.005

Repeat accuracy（*1 *2）

Uniformity（*1 *3）■Operating environment
CPUMemory*1*2*3〜 800,000 cd/㎡

Luminance±0.5％

Chromaticity±0.003

Luminance±2％

Chromaticity±0.003

・Camera unit
・ 3.0 m USB cable・USB memory key(Installation Software, andManual)・Carrying case

Based on white point of the white standard light source D65 of RGB backlight at the time of pre-shipment calibration.
Value in 10 iterations.
Value at 81 points within 90% range of camera image.

Intel Core i3 2.0GHz or higher
2GB or more

At least 300MB of free physical memory is needed at program execution.

HDDMore than 500MB of free space is needed when installing.

OSWindows 7 32bit/64bit

USB

Support USB3.0 and 2.0. Recommended 3.0 port if available.
Recommended to connect to USB controller with built-in Intel chipset.

MonitorEnvironmental condition

Windows 8 32bit/64bit

Windows 10 32bit/64bit

XGA（1024x768）or better resolution
Recommended SXGA（1280x1024）or better resolution with bigger than 17 inches display
0〜35℃（20〜80％RH※Windows 7,Windows8,Windows10, Excel are the registeredtrademarks or trademarks of Microsoft corporation in the U.S.
and other countries.
※Intel Core i3 is a trademark or registered trademark in the U.S.
and other countries of INTEL Corporation or its subsidiarycompanies.
※Other listed company or product names are the trademarks orregistered trademarks of their respective companies.
※Specifications as set forth may be modified without prior notice
due to improvement etc.

with no condensation）
Distribution source, Contact

MIYAKAWA CORPORATION
E-mail : ei4@mcp-hybec.co.jp
6-4 Akashicho, Chuo-ku, Tokyo 104-0044 JAPAN
Phone: +81-3-3543-8491 FAX : +81-3-3545-9219
http://www.mcp-hybec.co.jp/

Manufacturer

TOHOKU DENSHI CO., LTD.
2017-02 MCPCL300R0