Duct Inserted Sensors

DSD

Temperature

Humidity

CO2 Concentration

DSD-101-CO2-DSD-200-

Temperatureand Humidity

■ Sensor Type
Measurement
TargetTemperature
Humidity
Temperature,
Humidity
CO2Concentration

Measurement
Range-25 to 70
5 to 95 %RH-25 to 70
5 to 95 %RH
0 to 2000 ppm
0 to 5000 ppm■ Specifications
OutputPower Supply
Accuracy

Response
Characteristics

Dimensions
Mounting

Power
Consumption
Operating
Environment

Storage
Environment

WeightAccessories

OutputPt100 Class A 3-wire type
4 to 20 mA DC
Temperature: Pt100 Class A 3-wire type
Humidity: 4 to 20 mA DC
4 to 20 mA DC
1 to 5 V DC
4 to 20 mA DC
1 to 5 V DC

DSD-200-TR
Pt100 Class A
3-wire type

Model
DSD-200-TR
DSD-200-HA
DSD-200-TRHA
DSD-101-CO2-1A
DSD-101-CO2-1V
DSD-101-CO2-2A
DSD-101-CO2-2V

DSD-200-HA
DSD-200-TRHA
DSD-101-CO24 to 20 mA DC:
Temp: Pt100 Class A 4 to 20 mA DC:(Linear conversion
3-wire typeMax allowable load:corresponding toHumi: 4 to 20 mA DC
500
max.
0 to 100 %RH)
(Linear conversion 1 to 5 V DC:
Max allowable load: corresponding toMax allowable load:
0 to 100%RH)
1 k max.
100 k min.
Max allowable load: (Must be specified.)
1 k max.
Not required
24 V AC/DC 20% 50/60 Hz
24 V DC 10%
1.0 (-25 to 75 ) Within 5 %RH
Temp: 1.030 ppm
However,(at 23 )(-25 to 75 )
( 5% of indicated0.3 (-10 to 50 )However,value)
0.3 (-10 to 50 )Humi: Within
5 %RH (at 23 )Approx. 30 seconds Approx. 20 seconds Temp: Approx.30 sec Within 1 minute(at 2 m/s wind(at 2 m/s wind(at 2 m/s wind
(at 1.5 m/s windvelocity)velocity)velocity)
velocity)
Humi: Approx.20 sec(at 2 m/s windvelocity)
W86 x H86 x D293 mm
W100 x H133 x
D260 mm
Adjustable flange
Duct inserted, and fixed with the resin bracket.
DSDF-SQ80
recommended(sold separately)
Approx. 1 VA (for 24 V AC)
18 W
Approx. 1 W (for 24 V DC)
Temp: -25 to 70
Temp: 0 to 50
Humi: Less than 95 %RH (non-condensing)
Humi: 5 to 95 %RH(Operating environment temperature:(non-condensing)
Up to 50 : Less than 95 %RH,
50 to 70 : Less than 70 %RH)
Temp: -25 to 70
Temp: -20 to 60
Humi: Less than 95 %RH (non-condensing)
Humi: 5 to 95 %RH(Operating environment temperature:(non-condensing)
Up to 50 : Less than 95 %RH,
50 to 70 : Less than 70 %RH)
Approx.170 g (Including resin bracket))Approx. 190 g
Approx. 400 g(Including resin bracket)
Mounting screw: 4Mounting screw: 4 pieces,
Instruction manual:pieces,Resin bracket: 1 piece,
1 copyResin bracket: 1 piece, Packing: 1 piece,
Current/Voltage conversion resistor (RESPacking: 1 piece,S06-250): 1 piece,
Cable gland (OACable gland (OA-W1608E): 1 piece,W1608E): 1 piece,
Instruction manual: Instruction manual: 1 copy
1 copyAbbreviations: Temp: Temperature,

Humi: Humidity

■ Terminal Arrangement
DSD-200-TR
A
Temp.

Bb

A

DDC
PLC
Controller
RecorderBb

DSD-200-HA

DDC
PLC
Controller
RecorderHumid.

24V Gnd Out

24V
24 V AC/DCPower0V

When using 1 to 5 V DC output, connect the provided
Current/Voltage conversion resistor (RES-S06-250)
between Out – Gnd.

+DSD-200-TRHA

Humidity

B-b+

Temperature
DDCPLCController
Recorder

Temp.

DDC
PLC
Controller
RecorderHumid.

24V Gnd Out A

24V
24 V AC/DCPower0V

When using 1 to 5 V DC output, connect the provided
Current/Voltage conversion resistor (RES-S06-250)
between Out – Gnd.

ABb
Pt1000(Pt100)

DSD-101-CO2-

OUT
POWER

OUT:
1 to 5 V DC or 4 to 20 mA DC output (Must be specified.)
POWER: Power supply voltage 24 V DC
[Note] Do not reverse polarity.

■ External Dimensions (Scale: mm)
DSD-200-86293

68250
228 max.

4-φ54-φ1465Packing

4256φ14205642φ22

Resin bracket

5086

φ14G1/2

Resin bracket

(33)

100

φ30

DSD-101-CO2-Compatible electric wire
diameter 6 to 10

100

10060200SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel:+81-72-727-6100

Fax:+81-72-727-7006

URL:

http://www.shinko-technos.co.jp/e/E-mail:overseas@shinko-technos.co.jp

201908 E02