OEM’s #1 Choice for X-Ray
Sources and Generators
Focus on Performance,
Reliability, Form Factor,
and Versatility
www.vjtxray.com

Applications
Security, Industrial NDT, Food & Pharmaceutical, Quality & Safety, Electronics, Medical & Life Sciences,
Sorting & Recycling

Your Single Source
for X-Ray Sources
The IXS Series and the HVG series are highly reliable productlines from VJ X-ray. These series of products compriseof integrated X-ray Sources and stand alone high voltage X-ray
generators. Unique in design, these generators are exceptionallycompact and robust units that have superior performance.
The IXS series are integrated X-ray sources which incorporate ahigh voltage inverter, a filament supply, an X-ray tube, and a heatdissipation component into a single unit. The versatile platform
operates from 20 kV to 200 kV at up to 1000 Watts output power
continuously, and even higher in pulsing applications. The sources
can be controlled via user-friendly RS232 digital interface or analog
signals.
The HVG series are high voltage generators uniquely offer highstability and performance while retaining an extraordinarily
compact design. Either Bi-polar or Uni-polar, these units havea variety of kV and mA outputs and are capable of a wide arrayof applications. The High Voltage generators are designed to becontrolled by Ethernet, RS 232, or Analog.

Capable of operating under an extensive range of conditions,the IXS and HVG Series are ideal for many applications. These
include:•Non-destructive Testing•

Security Inspections: explosive detection, baggage 		
inspections, body screening•

Foreign Object Detection: food and beverage, health and 		
beauty, pharmaceuticals•

Medical: mammography, dental CT, bone densitometry, 		
medical C-arm•

Material Analysis: thickness gauging, X-ray fluorescence 		
(XRF)

•

Electronic Component Inspections•

Product Quality Inspections and Packaging Inspections

The IXS and HVG Series generators are cost-effective andreliable no matter what the application is. Their small form factor,
efficient heat dissipation, and leak-free tank make integrating
and maintaining these units simple and hassle-free.
In addition, the X-ray generators are backed by the support ofa technical team that has over 20 years of high voltage and X-ray
experience, and by VJ Technologies, a company that hasdelivered practical customized X-ray solutions for the pastquarter century.

Multipurpose X-Ray
Sources & Generators
Contents1.

IXS Series - Integrated X-ray Sources

IXS041k Beryllium Window		
8
IXS0505 Beryllium Window
10
		
IXS0520					12
		IXS0803					14			
IXS0808 				16
		IXS0808 Mini-Focus			18
		IXS1010					20
		IXS1015					22				
IXS1020 Water Cooled 			24
		
IXS1050 					26
		IXS1203 Mini-Focus		28
		
IXS1212					30
		IXS1620					32
		IXS1650					34
		IXS1650 Water Cooled			36				
IXS1680 					38
		IXS2050					402.

HVG Series - High Voltage Generators

HVG050					44
		HVG100					46
		HVG160 - 450				48
		3.

Accessories & Options

Accessories 				52		
				

1000

IXS Series - Integrated X-Ray Sources

500

800

ixs041k

200

IXS1050

IXS0520150Power (w)

IXS1020 Water-Cooled80

IXS1015

IXS101050

IXs0808,
IXS0808 Mini-Focus

IXS0505 Be-Window0

ixs0803

50kV

80kV

100kV

Multipurpose X-Ray
Sources & Generators

IXS1680

IXS1650 Water-Cooled, IXS1650

IXS2050

IXS1620

IXS1212IXS1203 Mini-Focus

120kV					160kV					200kV

IXS Series
Integrated X-Ray Sources
The IXS Series of Integrated sources incorporates the highvoltage generator, X-ray tube, and control electronics into single
compact products. These units boast high stability and performance over an expansive range of voltage and current. Versatile
design allows customization based on application need, andseamless integration into OEM systems.
Specification Range••••

Output Voltage:
Output Current:
Output Power:
Focal Spot Size:

10–200 kV
0.05–40.0 mA
5–1000 W
35μm–1.2 mm

Typical Applications
•
•
•
•
•
•
•
•

Security Scans
Industrial NDTFood & Pharmaceutical Quality Control
Electronics Inspections & Component Counting
Medical Imaging
Thickness Gauging
Research
Sorting & Separation			IXS041k Beryllium Window
40 kV, 1000 W
Multipurpose Sources
Applications

Food Inspection, Industrial NDTKey Features• Integrated High Voltage Generator, X-ray tube,
and Control Electronics• Be-Window for Low kV, Soft X-ray Applications• High Power with Water Cooling• Radiation Shielded• User Friendly RS232 Digital Interface
Specifications

Fan Beam X-ray Generator

Control Unit

Input Line Range 		

230 VAC, 50/60 Hz

Output kV 		

20 - 40 kV

Output mA 		

1.0 - 25.0 mA

Output Power 		

1000 W maximum continuous

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

X-ray Tube

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

Beam Port		
Fan beam of 60º max. (Cone Beam 					available upon request)

Ripple 						kV: ± 0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Operating Temperature

Dimensions

Generator: 250mm x 297mm x 125mm

		Control unit: 254mm x 247mm x 70mm

Weight			Generator: 14 kg			Control unit: 3 kg

X-ray Tube Type		Glass with 0.8 Be-Window

X-ray Focal Spot Size

1.3 mm as per IEC60336

Operating Environment

Storage Temperature
Repeatability		kV: <0.5% 				Thermal Cut Off					mA: <0.5%
Overshoot		 kV: ≤5% of rated output
Output Rise Time			kV Rise Time <1 sec. from 10% to 90%of the output voltage.		

Cooling			Water Cooled

Radiation shield 					Less 0.1mR/hr at 15cm from the 		surface of the chassis.

5°C to 35°C
-20°C to 55°C
60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

CoolerSafety and Regulatory
Designed to meet CE, CSA, TUV,
Compliances
EN60950 and EN61010A-1.		
																								

8Generator

LED Indicators

59+0.5F.S.

63.5+0.5
F.S.

205
64.5+0.5
F.S.

70
264.7
357.2
105

192

OP

Over Power fault

OCOver current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60±3ºC.

OVOver voltage fault

X-ray On

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

J1: AC Input
Unit: mmControl Unit

Pin Out

Name

N

NeutralGND

GroundL

230 VAC Input

J2: Interlock

254
247
70

Unit: mm

Graphical User Interface

Pin Out

Name

1

Interlock out

2

Interlock In

3

X-ray on relay contact common

4

X-ray on relay contact N/C

5

X-ray on Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

J3: RS232
Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

9

IXS0505 Beryllium Window
50 kV, 50 W
Sub-Compact Multipurpose Sources
Applications
XRF, Thickness Gauging, Food Inspection

Key Features
•
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, and Control 		Electronics
BE-Window for Low kV Applications
High Resolution
Compact and Robust
Radiation Shielded
User Friendly RS232 Digital Interface

Specifications

Cone Beam X-ray Generator

Control Unit

Input Line Range 		

90–264 VAC, 50/60 Hz

Output Voltage 		

10–50 kV

Output Current 		

1.0 mA max

Output Power 		

50 W maximum

Voltage Regulation
			
			
			

Load:<0.1% at max kV output over
the output mA range
Line:<0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load:<0.1% at max mA output over
the output kV range
Line:<0.1% at max mA output overthe input line range

X-ray			GlassX-ray Focal Spot Size

165 µm maximum

Beam Port		

Cone beam 25º

Ripple 						

kV:<0.5% p-p of maximum output		mA: <20µA% p-p of maximum output

Operating Environment

Generator: 203mm x 125mm x 70mm
			Control unit: 254mm x 134mm x 65mm

Dimensions 		

Weight						
X-ray Tube

Operating Temperature
Repeatability		kV:<0.5% 							mA:<0.5%
Storage Temperature
Stability			kV: 0.2% over 4 hours

Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤1 sec. from
10%–90% of max rated output

Cooling			

External Air Cooling required

Generator: 7 Kg
Control unit: 2 Kg

Thermal Cut Off 		

5°C to 40°C
-20°C to 85°C
60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

Designed to meet CE, CSA TUVCompliances

EN60950 and EN61010A-1

10LED Indicators

Generator
70203
111.5+0.8
F.S.
35+0.8F.S.

125 138

8

7

X-ray On

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

OV

Over voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

5

6

4

3

2

1

J1: Interlock Connection/ X-ray On Relay

23˚

D

Unit: mmControl Unit
225254

C

B

254

225

134

A

Unit: mm

Graphical User Interface

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

J2: RS232 Interface

65

8

Pin Out7

6

Pin Out

Name

1

N/A

2

TX-

3
4

RX+
5

N/A
4

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

3

2

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

111

IXS0520
50 kV, 200 W
Small Focus Multipurpose Sources
Applications
Baggage Inspections, Security Body Scanners, Food Safety, Industrial NDTKey Features
•
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, and Control
Electronics
Compact and RobustHigh Stability
High Resolution: 0.4 mm Focal Spot
Radiation Shielded
User Friendly RS232 Digital Interface

Specifications

X-ray Generator

Input Line Range 		

120/240 VAC, 50/60 Hz

Output kV 					

20–50 kV					*up to 80 kV also available
Safety and Regulatory

Output mA 		

0.5–4.0 mA

Compliances

Output Power 					

80 W continuous/ 200 W peak 		
30% or less Duty Cycle

Dimensions

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

Ripple 						kV: <1.0% p-p of maximum output		mA: <0.5% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: ≤5% of rated output
Output Rise Time			

Rise Time ≤300 ms from
10%–90% of max rated output

Cooling			

Air Cooled

Control Unit

Designed to meet CE, CSA, TUV,
EN60950, and IEC/EN61010-1
Generator: 200mm x 184mm x 134mm

		Control unit: 254mm x 134mm x 65mm

Weight			Generator: 11 Kg

			Control unit: 2 Kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size

0.4 - 0.8 mm as per IEC60336

Beam Port				

Fan beam: 80º X 10º , 74º X 12º				Cone beam: 30º

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40

12

Generator

LED Indicators

200

134

74˚

184

8

12˚

7

44+0.8F.S.

Unit: mm

OVOver voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC
4

Pin Out

Name

1

Interlock out

2
7

5

6

4

2

3

225254CD

B225254

C

225

65

A

1

Interlock in
1

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

D

C

134

4

3

Name

1

N/A

2

TX-

4

6

Graphical User Interface
65

5

Pin OutB

3

Unit: mm
7

8

6

2

J2: RS232 Interface

B

7

3

254

134

8

Illuminated when power is present

J1: Interlock Connection/ X-ray On Relay

100+0.8
F.S.

A

Power

5

D

8

Illuminated when interlock
is closed and HV is enabled

685.7+0.8
F.S.

Control Unit

X-ray On

2

RX+
N/A
4

5

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

3

2

A

1

N/A

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

131

IXS0803
80 kV, 30 W
Multipurpose Sources
Applications
Food & Pharmaceutical Inspections, Fill Level Check

Key Features• Integrated High Voltage Generator, X-ray Tube, and Control
Electronics• Radiation Shielded• User Friendly RS232 Digital Interface• 24 VDC Input
Specifications
Input Line Range 		

24VDC ± 10%

Output kV 		

20 - 80 kV

Output mA 		

0 - 1.0 mA

Output Power 		

30 W continuous maximum

Voltage Regulation
			
			
			

Load: <0.2% at max kV output over
the output mA range
Line: <0.2% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

Ripple 						kV: < ± 0.5% rms of maximum output
mA: < ± 0.5% rms of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤1 sec to within 		
1% of programmed kV		

Cooling			

Air Cooled			

Generator view (front)

Control unit view (rear)

Safety and Regulatory

Designed to meet CE, and IEC/EN61010-1

Dimensions

Generator: 198 mm x 125 mm x 75 mm
		Control unit: 176 mm x 125 mm x 38 mm

Weight			Generator: 7 kg			Control unit: 1 kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size

0.8 mm as per IEC60336

Beam Port		

Cone beam: 25º

Operating Environment
Operating Temperature

5°C to 40°C

Storage Temperature-20°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40		

14

Generator

LED Indicators
39

114.3

125

100+0.8
F.S.

Power

Illuminated when power is present

X-ray On

Illuminated when interlock
is closed and HV is enabled

Arc

ARC-ing fault

OCOver current fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

OP

Illuminated when selected power
exceeds the rated power

OV

Over voltage fault

25˚

J1: Interlock Connection/ X-ray On Relay
8.5+0.8
F.S.

176

198

68+0.8F.S.

Unit: mm

Pin Out

Name

1

Interlock 1 in

2Interlock 1 in

3Interlock 1 out

4

N/A

5

Interlock 2 in

6Interlock 1 in

7

Interlock 1 out

8Interlock 2 out

9

X-ray Enable (TBD)

J2: RS232 Interface

Graphical User Interface

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6External kV program

7

External mA program

8

External kV monitor

9External mA monitor

J3: Power Input
L

G+24 VDC

Ground15

IXS0808
80 kV, 80 W
Multipurpose Sources
Applications
Food Inspection Systems, Security Scanners, Industrial NDT, Product Quality Monitoring

Key Features
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, and Control 		Electronics
Water Cooled Option for Improved Stability
Compact and Robust
Radiation Shielded
User Friendly RS232 Digital Interface

Fan Beam X-ray Generator

Control Unit

Safety and Regulatory

Designed to meet CE, CSA, TUVCompliances

EN60950 and EN61010A-1

Specifications
Input Line Range 		

90–264 VAC, 50/60 Hz

Output kV 		

20–80 kV

Output mA		

0.05 - 1.0 mA

Dimensions 		

Generator: 225mm x 144mm x 134mm

Control unit: 254mm x 134mm x 65mm
Generator: 11 Kg
Control unit: 2 Kg

Output Power 		

80 W continuous maximum			

Voltage Regulation
			
			
			

Load: <0.2% at max kV output over
the output mA range
Line: <0.2% at max kV output overthe input line range

Weight						

Current Regulation
			
			
			

Load: <0.1% at max mA output over
the output kV range
Line: <0.1% at max mA output overthe input line range

X-ray Tube Type		

Glass

X-ray Focal Spot Size

0.8 mm as per IEC60336

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Stability						kV: <0.01% per °C over the 			operational ambient temperature range

Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤500 msec. from
10%–90% of max rated output

Cooling						

Air Cooled 				(Water cooled option also available)

X-ray Tube

Beam Port		
Fan beam: 80º X 10º max
			Cone beam: 30º
Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off 		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40

16

LED Indicators

Generator

219.7
144

80˚

116±0.8
F.S.

46±0.8
F.S.

225
7

8

134

6

D

Unit: mmControl Unit
225254

C

B

225

Power

Illuminated when power is present

OV

Over voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

5

4

3

2

1

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

C

254

134

J2: RS232 Interface

65

A

8

Graphical User Interface

Illuminated when interlock
is closed and HV is enabled

J1: Interlock Connection/ X-ray On Relay

248

195

X-ray On

7

Unit: mm6

Pin Out

Name

1

N/A

2

TX-

3

RX+

5

4

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

4

N/A

3

2

1

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

17

IXS0808 Mini-Focus
80 kV, 56 W
Mini-Focus Sources
Applications
Electronic Inspections, Food & Pharmaceutical Safety Inspections, Industrial NDTKey Features•

Integrated High Voltage Generator, X-ray Tube, and Control 		Electronics

•
•
•
•
•
•

Wide Beam Angle
High Resolution
Compact and Robust
Cone or Fan beam available for flat panel or line sensor detection
Radiation Shielded
User Friendly RS232 Digital Interface

Specifications
Input Line Range 		

90–264 VAC, 50/60 Hz

Output Voltage 		

20–80 kV

Output Current 		

0.2 - 0.7 mA

Output Power 		

56 W continuous maximum

Voltage Regulation
			
			
			

Load:<0.1% at max kV output over
the output mA range
Line:<0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load:<0.1% at max mA output over
the output kV range
Line:<0.1% at max mA output overthe input line range

Ripple 						kV: <0.5% p-p of maximum output		mA: <20µA p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Stability						kV: <0.01% per °C over the 			operational ambient temperature range

Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤500 msec. from
10%–90% of max rated output

Cooling			

External Air Cooling required

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

Designed to meet CE, CSA, TUVCompact Configuration shown with
Cone Beam and Integrated Control Unit

Compliances

Control Unit

EN60950 and EN61010A-1

Dimensions 		

Generator: 225mm x 144mm x 134mm

			Control unit: 254mm x 134mm x 65mm

Weight						Generator: 11 Kg
Control unit: 2 Kg

X-ray Tube
X-ray			GlassX-ray Focal Spot Size

35 - 50 µm

Beam Port		
Fan beam: 80º X 10º max
			Cone beam: 30º
Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off 		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

18

LED Indicators

Generator

219.7

X-ray On

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

OV

Over voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

144

80˚

116±0.8
F.S.

46±0.8
F.S.

225
8

134

J1: Interlock Connection/ X-ray On Relay

6

7

248

195

Unit: mm

D

Control Unit
225254C

5

4

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

3

2

1

B

225

254

J2: RS232 Interface

134
65

A

8

Graphical User Interface

7

Unit: mm6

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

4

3

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

2

1

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

19

IXS1010
100 kV, 100 W
Multipurpose Sources
Applications
Thickness Gauging, X-ray Analysis

Key Features• Integrated High Voltage Generator, X-ray Tube, and Control

•
•
•
•
•

Electronics
High Stability
Form Factor Designed Specifically for Thickness Gauging
Radiation Shielded
User Friendly RS232 Digital Interface
24 VDC Input

Input Line Range 		

24 VDC ± 10%

Output kV 		

30 - 100 kV

Output mA 		

0.05 - 2.0 mA

Output Power 		

100 W (Continuous)

Generator with mounted control unit Generator with 40º beam port

Radiation shield 		
Less than 0.5 mR/hr at 5cm from the 					surface of the chassis as per FDA 21 					CFR 1020.40

Voltage Regulation
Load: <0.1% at max kV output over		
Safety and Regulatory
Designed to meet CE, CSA, TUV 					the output mA range						EN60950
and EN61010A-1
			
Line: <0.1% at max kV output over					the input line

Dimensions

Current Regulation
Load: <0.1% at max mA output over					the output kV range				Weight						Line: <0.1% at max mA output over					the input line range load:
Ripple 						kV: < 0.5% rms of maximum output		mA: < 20 µA p-p of maximum output

Repeatability			

kV: <0.5% 				mA: <0.5%

kV Stability					

0.01% per °C over the 			operational ambient temperature range;			

0.1% in 8 hours after 40 min warm up

Overshoot		 kV: <5% of rated output

			Standard Rise Time <1 sec to within 		
1% of programmed value

Cooling						Water Cooled 				(Optional air cool for lower output units)

Output Rise Time 		

Generator: 156mm x 350mm x 174mm
Generator: ~15 kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size

0.8 mm as per IEC60336

Beam Port		

Cone beam: 40º

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-20°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

20LED Indicators

Generator
140

POWER

Illuminated when Power is present

X-ray ON

Illuminated when Interlock is closed
and HV is enabled

ARC

ARC-ing fault

OCOver Current Fault

OT

Illuminated when oil temperature
exceeds 60±3ºC

OP

Illuminated when selected power
exceeds the rated power

OV

Over Voltage fault

38

79+0.8F.S.

59.5+0.8
F.S.

106.9+0.8
F.S.

40˚

J1 Connector: (Interlock 9 Pin Male)

351

349

444.1

156.2

Unit: mm172

Graphical User Interface

Pin Out

Name

1

Interlock In

2

Interlock In

3Interlock Out

4

X-ray On Relay: On=GND, Off=Open

5

N/A

6

N/A

7

Interlock Out

8

X-ray Enable Out

9

X-ray Enable In

J2 Connector: (RS232 9 Pin Female)
Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6External kV Program

7

External uA Program

8

External kV Monitor

9External uA Monitor

J3 Connector
L

G+24V DC

Ground21

IXS1015
100 kV, 150 W
Multipurpose Sources
Applications
Food Inspection Systems, Security Scanners, Industrial NDT, Product Quality Monitoring

Key Features
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, and Control 		Electronics
High Stability
Compact and Robust
Radiation Shielded
User Friendly RS232 Digital Interface

Specifications
Input Line Range 		

90–264 VAC, 50/60 Hz

Output kV		

25–100 kV

Output mA 		

0.05–6.0 mA

Output Power 		

150 W continuous maximum

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.1% at max mA output over
the output kV range
Line: <0.1% at max mA output overthe input line range

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: ≤ 5% of rated output
Output Rise Time			Standard Rise Time ≤500 ms from
10%–90% of max rated output

Cooling			
Air Cooled (Water cooled option also 					available)

Control Unit

Cone Beam X-ray Generator
Dimensions 		

Generator: 254mm x 210mm x 143mm

			Control unit: 254mm x 134mm x 65mm

Weight			Generator: 14 Kg

			Control unit: 2 Kg

X-ray Tube		Control Unit

X-ray Tube Type		

Glass

X-ray Focal Spot Size		

0.8 mm as per IEC60336
0.5 mm available upon request

Beam Port		
Fan beam: 80º X 10º max
			Cone beam: 30º
Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off 		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

Designed to meet CE, CSA, TUVCompliances

EN60950 and IEC/EN61010-1

22		Generator

LED Indicators

210

143
60.1+0.8
F.S.

254

30˚
3.4+0.8
F.S.
8

90.7+0.8
F.S.

7

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

OV

Over voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

5

6

4

3

2

1

J1: Interlock Connection/ X-ray On Relay

D

Unit: mmControl Unit
225254C

B

225

X-ray On254

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

134
J2: RS232 Interface

65

A

8

Graphical User Interface

7

6

Unit: mm

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

5

4

3

2

1

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

23

IXS1020 Water Cooled
100 kV, 200 W
Multipurpose Sources
Applications
Thickness Gauging, X-ray Analysis, Industrial NDT, Security Scanners

Key Features
•
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube,Control Electronics
and Liquid to Liquid Heat Exchanger
High Stability
Compact and Robust
Radiation Shielded
User Friendly RS232 Digital Interface
Ethernet & Wifi (optional)
Cone Beam X-ray Generator

Specifications
Input Line Range 		

90–264 VAC, 50/60 Hz

Output kV 		

25–100 kV

Output mA 		

0.05–8.0 mA

Output Power 		

200 W continuous maximum

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.1% at max mA output over
the output kV range
Line: <0.1% at max mA output overthe input line range

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Stability						kV: <0.01% per °C over the 			operational ambient temperature range

Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤500 ms from
10%–90% of max rated output

Cooling						Water Cooled 				(see accessories for cooler options)Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40

Optional Water to Air Cooler

Safety and Regulatory

Designed to meet CE, CSA, TUVCompliances

EN60950 and IEC/EN61010-1

Dimensions 		

Generator: 279mm x 210mm x 143mm

			Control unit: 254mm x 134mm x 65mm

Weight			Generator: 14 Kg

			Control unit: 2 Kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size			

0.8 mm as per IEC60336
0.5 mm available upon request

Beam Port		
Fan beam: 80º X 10º max
			Cone beam: 30º
Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off 		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

24Generator

LED Indicators

143

210
60.1±0.8
F.S.

90.7±0.8
F.S.

150.6±0.8
F.S.

279 25430˚

8Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

OV

Over voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

5

6

7

X-ray On

4

3

2

1

J1: Interlock Connection/ X-ray On Relay

374

Unit: mm

D

Control Unit
225254CD

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

C

B

225

B

254

134

J2: RS232 Interface

65

A

8

Graphical User Interface

7

Unit: mm

6

5

Pin Out

Name

1

N/A

2

TX-

3

4

RX+

A

3

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

2

1

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

25

IXS1050
100 kV, 500 W
Multipurpose Sources
Applications
Dental CT, Panoramic Dental, Medical Research

Key Features• Ideal for panoramic dental and CT applications• Integrated High Voltage Generator, X-ray Tube, and Control
Electronics• Radiation Shielded• User Friendly RS232 Digital Interface
Specifications
Input Line Range 		

230 VAC, 50/60 Hz

Output kV 		

40 - 100 kV

Output mA 		

2.0 - 10.0 mA

Output Power 		
150 W continuous maximum
			500W peak power						* up to 1 kW peak also available
Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Stability						kV: ± 1.0% 				mA: ± 1.0%

Overshoot		 kV: ≤5% of rated output

Cone Beam X-ray Generator

Control Unit

Generator: 254mm x 192mm x 143mm

Dimensions
		Control unit: 254mm x 247mm x 70mm

Weight			Generator: 12 kg			Control unit: 3 kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size

0.4 mm as per IEC60336

Beam Port		

Cone beam: 30º

Operating Environment
Operating Temperature

5°C to 40°C

Storage Temperature-20°C to 80°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Output Rise TimekV Rise Time <200 msec to within 1%
			of selected value			
Cooling			

Air Cooled			

Radiation shield 					Less than 1 mGy/hr at 1 m from the 		surface of the chassis

Safety and Regulatory
Compliances

Designed to meet CE, CSA, TUV,
IEC/EN60601-1 			

26

LED Indicators

Generator
192

143

150.4±0.8
F.S.

254

30˚

OP

Over Power fault

OCOver current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60±3ºC.

OVOver voltage fault

X-ray On

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

329

J1: AC Input
55.9±0.8
F.S.

52±0.8F.S.

Unit: mmN

NeutralGND

GroundL

230 VAC Input

Control Unit
J2: Interlock

254

247

70

Unit: mm

Graphical User Interface

Pin Out

Name

1

Interlock out

2

Interlock In

3

X-ray on relay contact common

4

X-ray on relay contact N/C

5

X-ray on Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

J3: RS232 Interface
Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

27

IXS1203 Mini-Focus
120 kV, 36 W
Mini-Focus Sources
Applications

		Electronic Inspections, Food & Pharmaceutical Safety Inspections, Industrial NDTKey Features
•
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, and Control 		Electronics
High Resolution
Compact and Robust
Cone or Fan beam available for flat panel or line sensor detection
Radiation Shielded
User Friendly RS232 Digital Interface

Cone Beam X-ray Generator

Specifications

Control Unit

Input Line Range 		

115–230 VAC, 50/60 Hz

Output Voltage 		

40–120 kV

Output Current 		

0.05 - 0.3 mA

Output Power 		

36 W continuous maximum

			Control unit: 254mm x 247mm x 70mm

Voltage Regulation
			
			
			

Load:<0.1% at max kV output over
the output mA range
Line:<0.1% at max kV output overthe input line range

Weight			Generator: 23 kg			Control unit: 3 kg

Current Regulation
			
			
			

Load:<0.5% at max mA output over
the output kV range
Line:<0.5% at max mA output overthe input line range

X-ray			Ripple 						kV: <0.5% p-p of maximum output		mA: <20µA p-p of maximum output

Beam Port					

Repeatability		kV: <0.5% 							mA: <0.5%
Stability						kV: <0.01% per °C over the 			operational ambient temperature range

Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤500 msec. from
10%–90% of max rated output

Cooling			

External Air Cooling required

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

Designed to meet CE, CSA, TUVCompliances

EN60950 and EN61010A-1

Dimensions 		

Generator: 417mm x 393mm x 143mm

X-ray Tube
Glass

X-ray Focal Spot Size
50 µm						
Fan beam: 80º X 10º max
Cone beam: 30º, 40º

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-20°C to 85°C

Thermal Cut Off 		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

28

Generator

LED Indicators

408.3

143

349

80˚

392.5

40˚

X-ray On

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

OV

Over voltage fault

OP

Illuminated when selected power
exceeds the rated power

OC

Over current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

172.8±0.8
F.S.

247
174.5±0.8
F.S.

J1: Interlock Connection/ X-ray On Relay

60.1±0.8
F.S.

70

Control Unit

254
247

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

J2: RS232 Interface
70

Graphical User Interface
254

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

J3: AC Input
N

NeutralGND

GroundL

90-264 VAC Input

29

IXS1212 Portable
120 kV, 120 W
Portable X-Ray Source
Applications

Non-Destructive Testing, Security, General X-ray Operations

Key Features• Battery Operated• Integrated High Voltage Generator, X-ray Tube,
Control Electronics, and Exchangeable Lithium Battery Pack
• Radiation Shielded• Wifi, and Ethernet or RS232• Accessories: Tripod Mount, Protective Cover,		and Carrying Handle
Generator shown with protective
Input Line Range 		
24VDC ± 10%, or 				cover and handle
			24V LiFePO4 Battery Pack

Generator shown mounted
on tripod

Output kV 		

30–120 kV

Output mA 		

0.2–1.0 mA

Cooling			Battery Power 					Continuous: 14min @120W 			
Pulsing: 21min, 15sec On/15sec Off

Radiation Shield 		
<6mR/hr at 5cm from the surface with			beam port blocked.		

Battery Charging Time

2 hours from low line (21V)

Safety and Regulatory

Voltage Regulation
Load: < 0.2% for mA changes over 		Dimensions
			
specified range (with constant ambient			temperature and line input.)			
Weight			
Line: < 0.2% for Line Input changes 					over specified range (with constant 		
X-ray Tube			ambient temperature and load).
X-ray Tube Type		
Current Regulation
Load:< 0.5% for mA changes over 		
X-ray Focal Spot Size			
specified kV range (with constant 		
			ambient temperature and line input).
Beam Port					
Line: < 0.5% for Line Input changes 								over specified range (with constant 		
ambient temperature and load).

Ripple 						kV: < ± 0.5% rms of maximum output
mA: < ± 0.5% rms of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: <5% of rated output
Output Rise Time			Standard Rise Time ≤0.5 Sec. to within
1% of programmed kV		Air Cooled			

Designed to meet CE, and IEC/EN61010-1
Generator: 245mm x 80mm x 184mm
5.9 kg (Generator only)Glass
0.5 mm as per IEC60336
Cone beam: 35º

Operating Environment
Operating Temperature-20°C to 50°C

Storage Temperature-30°C to 80°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing
IP Rating		

IP60

30

Connectors

Generator80

44 1 F.S

184

245

8

Safety Key Lock

J2

RJ45 Connector (Ethernet or RS232)

J3
7

5

6

4

40°

2

3

.5°

Input Power and Battery Charger
117

130 ±1
F.S

20°
D

J1

INPUT POWER CONNECTOR

35°

D

4

5

3

C

C

Pin Out

Unit: mm

B

1

Graphical User Interface
PERMISSIBLE DEVIATIONS FOR RANGES OF PERMISSIBLE DEVIATIONS FOR LINEAR DIMENSIONSLENGTHS, IN MILLIMETERS,OF THE SHORTER WITHOUT TOLERANCE (EXCEPT FOR BROKEN EDGES)SIDE OF THE ANGLE CONCERNED.

AUP TO 10

± 1°

OVER 10 UP TO 50± 0°30'OVER 50 UP TO 120± 0°20'OVER 120 UP TO 400± 0°10'OVER 40087

6± 0°5'UP TO 3

± 0.1

OVER 3 UP TO 6

± 0.1

OVER 6 UP TO 30± 0.2

OVER 30 UP TO 120± 0.3

OVER 120 UP TO 400± 0.5

OVER 400 UP TO 1000

± 0.8

OVER 1000 UP TO 2000

± 1.2

OVER 2000 UP TO 4000

±2

5PROJECTION
TYPE:

3RD ANGLEPROPRIETARY AND CONFIDENTIAL
THE INFORMATION CONTAINED IN THISDRAWING IS THE SOLE PROPERTY OF
VJ TECHNOLOGIES. ANY REPRODUCTION
IN PART OR AS A WHOLE WITHOUT THE
WRITTEN PERMISSION OF VJ TECHNOLOGIESIS PROHIBITED.

AL6061-T6

DRAWN

FINISH:

R.C

GOLD ALODINEUSED ONCHECKEDR.C

APPROVEDR.C

QUANTITY

4

B

Auxiliary Power 24VDC/2W

4

24VDC Battery Charger

3

Battery Charger Return6B

10/10/2014

SCALE 1:5

24VDC Return
24VDC/10A External Power SupplyA

89 CARLOUGH RD, BOHEMIA,N.Y. 11716-2903

10/10/2014

73

PLATE, MULT SIDE, 100KV, SINGLE ENDED
SIZE DWG NO

1

2

V.J.TECHNOLOGIES

10/10/2014

2

Name

5

MATERIAL:

68

7DF3000-125M

8

1

SHEET

DOC. NO. ZT3000-001Rev. A

2REV
1 OF 1

Auxiliary Return
Interlock Out
Interlock In

1

Accessories

1
5

2
3

4

6

5

Item

Part Number

Description

1

AS3001-288M

Battery pack( inc. 24V LiFeO4 battery)

2

AS3001-300M

Charging Station

3

DS3000-041M

Protective Cover and Handle

4

DB3000-122M

Tripod Mount

5

DS3000-106

Laser Alighment Guide

6

JP3000-019

Carrying Case31

IXS1620
160 kV, 200 W
Multipurpose Sources
Applications
Medical Imaging, Security Inspections, Industrial NDT, Food Inspection, Quality Monitoring

Key Features
•
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, and Control
Electronics
Compact and Robust
Wide Beam Angle
Fast Rise time
Radiation Shielded
User Friendly RS232 Digital Interface

Input Line Range 		

115/230 VAC, 50/60 Hz

Output kV 		

25 - 160 kV

Output mA 		

0.2 - 8.0 mA

Output Power 					

200 W continuous 				Higher peak power with Duty Cycle

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: ≤5% of rated output
Output Rise Time			

Rise Time ≤200 ms from
10%–90% of max rated output

Cooling			

External Air Cooling required

Generator shown with
Integrated Control Unit		

Control Unit options:
160kV or 100kV

Control unit: 254mm x 247mm x 70mm

			(Smaller control unit for models 					≤100kV)
Weight			Generator: 21 Kg

			Control unit: 3 Kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size			

0.8 mm as per IEC60336			
0.5 mm available upon request

Beam Port				

Fan beam: 80º X 10º 			Cone beam: 30º, 40º

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

Designed to meet CE, CSA, TUV,

Compliances

EN60950, and IEC/EN61010-1

Dimensions

Generator: 419mm x 265mm x 143mm

32

LED Indicators

Generator
143

419

Power

Illuminated when power is present

349

X-ray On

Illuminated when interlock
is closed and HV is enabled

Arc

ARC-ing fault

OCOver current fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

OP

Illuminated when selected power
exceeds the rated power

OV

Over voltage fault

40˚

265

80˚
81.2±0.8
F.S.

60.1±0.8
F.S.

174.5±0.8
F.S.

247

Control Unit

J2: RS232 Interface
Unit: mm70

254

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

J3: 24 VDC Pump/Fan
24770

Unit: mm

Pin Out

Name++24 VDC-

Return++24 VDC-

ReturnJ4: Interlock Connection/ X-ray On Relay

Graphical User Interface
254

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

33

IXS1650
160 kV, 500 W
Multipurpose Sources
Applications
Industrial NDT, Security Scanners, Medical Research, Product Quality Monitoring

Key Features
•
•
•
•
•

Ideal Alternative to Conventional High Voltage Generator and 		
X-ray Tube Setup
Integrated High Voltage Generator, X-ray Tube, Control
Electronics, and Heat Radiator
Radiation Shielded
Cone or Fan beam available for flat panel or line sensor detection
User Friendly RS232 Digital Interface
Generator

Input Line Range 		

115/230 VAC, 50/60 Hz

Output kV		

25–160 kV

Output mA 		

0.2 – 8.0 mA

Output Power 		

500 W continuous maximum

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

Ripple 						kV: <1.0% p-p of maximum output		mA: <1.0% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: ≤5% of rated output

Control Unit
Generator: 417mm x 393mm x 143mm

Dimensions
		Control unit: 254mm x 247mm x 70mm

Weight			Generator: 23 Kg

			Control unit: 3 Kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size			

0.8 mm as per IEC60336
0.5 mm available upon request

Beam Port				

Fan beam: 80º X 10º 			Cone beam: 30º, 40º

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Output Rise Time						Standard Rise Time ≤ 500 msec. from
10%–90% of max rated output(Faster Rise Time available upon request)

Cooling						

Air Cooled				(Several heat exchanger options available)

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

Designed to meet CE, CSA, TUV,

Compliances

EN60950, and IEC/EN61010-1

34

Generator

LED Indicators

408.3

143

349

80˚

392.5

40˚

172.8±0.8
F.S.

Power

Illuminated when power is present

X-ray On

Illuminated when interlock
is closed and HV is enabled

Arc

ARC-ing fault

OCOver current fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

OP

Illuminated when selected power
exceeds the rated power

OV

Over voltage fault

J2: RS232 Interface
60.1±0.8
F.S.

247

174.5±0.8
F.S.

Unit: mm

70

Control Unit

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

254

J3: 24 VDC Pump/Fan

24770

Unit: mm

Pin Out

Name++24 VDC-

Return++24 VDC-

ReturnJ4: Interlock Connection/ X-ray On Relay

Graphical User Interface

254

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

35

IXS1650 Water Cooled
160 kV, 500 W
Multipurpose Sources
Applications
Industrial NDT, Thickness Gauging, Medical Research, Product Quality Monitoring

Key Features
•
•
•
•
•

Integrated High Voltage Generator, X-ray Tube, Control
Electronics, and Liquid to Liquid Heat Exchanger
Compact and Robust
Exceptional Stability
Radiation Shielded
User Friendly RS232 Digital Interface

Input Line Range 		

115/230 VAC, 50/60 Hz

Output kV 		

20–160 kV

Output mA 		

0.2–8.0 mA

Output Power 		

500 W continuous maximum

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Current Regulation
			
			
			

Load: <0.1% at max mA output over
the output kV range
Line: <0.1% at max mA output overthe input line range

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Repeatability		kV: <0.5% 							mA: <0.5%
Stability						kV: <0.01% per °C over the 			operational ambient temperature range

Overshoot		 kV: ≤5% of rated output
Output Rise Time			Standard Rise Time ≤500 ms from
10%–90% of max rated output

Cooling						Water Cooled 				(see accessories for cooler options)

Optional Water to Air Cooler(see accessories and options)

Generator

Generator: 419mm x 338mm x 143mm

Dimensions
		Control unit: 254mm x 247mm x 70mm

Weight			Generator: 23 Kg

			Control unit: 3 Kg

X-ray Tube
X-ray Tube Type		

Glass

X-ray Focal Spot Size

0.4–0.8 mm as per IEC60336

Beam Port				

Fan beam: 80º X 10º 			Cone beam: 30º ,40º

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory
Compliances

Designed to meet CE, CSA, TUV,
EN60950, and IEC/EN61010-1

36

LED Indicators

Generator
143

397

397334
334

45˚

307

90˚
96.2±0.8
F.S.

159.5
159.5±0.8
FF.S.S.

60.1±0.8
F.S.

10°
247

Power

Illuminated when power is present

X-ray On

Illuminated when interlock
is closed and HV is enabled

Arc

ARC-ing fault

OCOver current fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

OP

Illuminated when selected power
exceeds the rated power

OV

Over voltage fault

J2: RS232 Interface
Unit: mm

70

Control Unit254

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

J3: 24 VDC Pump/Fan

24770

Unit: mm

Pin Out

Name++24 VDC-

Return++24 VDC-

ReturnJ4: Interlock Connection/ X-ray On Relay

Graphical User Interface

254

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

37

IXS1680
160 kV, 800 W
Multipurpose Sources
Applications
Industrial NDT, Food Inspection, Security Scanners, Medical Research

Key Features
•
•
•
•
•
•

Ideal Alternative to Conventional High Voltage Generator and 		
X-ray Tube Setup
Integrated High Voltage Generator, X-ray Tube, Control
Electronics, and Heat Radiator
Radiation Shielded
Cone or Fan beam available for flat panel or line sensor detection
User Friendly RS232 Digital Interface
Ethernet & Wifi (optional)

Input Line Range 		

230 VAC, 50/60 Hz

Output kV 		

40–160 kV

Output mA 		

0.5–10.0 mA

Generator

Control Unit

Generator: 417mm x 409mm x 143mm

Dimensions
		Control unit: 254mm x 247mm x 70mm

Output Power 		

800 W continuous maximum

Voltage Regulation
			
			
			

Load: <0.1% at max kV output over
the output mA range
Line: <0.1% at max kV output overthe input line range

Weight			Generator: 23 Kg

			Control unit: 3 Kg

Current Regulation
			
			
			

Load: <0.5% at max mA output over
the output kV range
Line: <0.5% at max mA output overthe input line range

X-ray Tube Type		

Glass

X-ray Focal Spot Size

1.2 mm as per IEC60336

Ripple 						kV: <0.5% p-p of maximum output		mA: <0.5% p-p of maximum output

Beam Port				

Fan beam: 80º X 10º 			Cone beam: 30º, 40º

Repeatability		kV: <0.5% 							mA: <0.5%
Overshoot		 kV: ≤5% of rated output
Output Rise Time
Standard Rise Time ≤500 msec. from			
10%–90% of max rated output
			(Faster Rise Time available upon 					request)
Cooling						Compliances

Operating Environment
Operating Temperature

0°C to 40°C

Storage Temperature-40°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Humidity		 98% non-condensing

Air Cooled				(Water cooled option also available)

Radiation shield 		
Less than 0.5 mR/hr at 5 cm from the			surface of the chassis as per FDA 21 					CFR 1020.40
Safety and Regulatory

X-ray Tube

Designed to meet CE, CSA, TUV,
EN60950, and IEC/EN61010-1

38

Generator

LED Indicators

143

409

40˚
80˚
81.2±0.8
F.S.
174.5±0.8
F.S.

247

60.1±0.8
F.S.

349
Unit: mm

419.5
70Control Unit

254

Power

Illuminated when power is present

X-ray On

Illuminated when interlock
is closed and HV is enabled

Arc

ARC-ing fault

OCOver current fault

OT

Illuminated when oil temperature
exceeds 60ºC ± 3ºC

OP

Illuminated when selected power
exceeds the rated power

OV

Over voltage fault

J2: RS232 Interface
Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

J3: 24 VDC Pump/Fan
24770

Unit: mm

Pin Out

Name++24 VDC-

Return++24 VDC-

ReturnJ4: Interlock Connection/ X-ray On Relay

Graphical User Interface

254

Pin Out

Name

1

Interlock out

2

Interlock in

3

X-ray On Relay contact common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

39

IXS2050
200 kV, 500 W
Multipurpose Sources
Applications
Industrial NDT, Security, General X-ray Operations

Key Features• Ideal Alternative to Conventional High Voltage Generator and		
X-ray Tube Setup• Integrated High Voltage Generator, X-ray Tube,
Control Electronics, and Heat Radiator• Radiation Shielded• Large Beam Angle Suitable for Compact System Design• Cone or Fan Beam Available for Flat Panel Or Line Sensor Detection•

User Friendly RS232 Digital Interface

Input Line Range 		

230 VAC, 50/60 Hz

Output kV 		

80–200 kV

Output mA 		

0.2–6.0 mA

Output Power 		

500 W continuous maximum

Fan Beam X-ray Generator

Control Box

Safety and Regulatory
Designed to meet CE, CSA, TUV, 					EN60950, and IEC/EN61010-1
Dimensions

Voltage Regulation
Load: <0.1% at max kV output over					the output mA range			
Weight						Line: <0.1% at max kV output over					the input line range

Generator: 436mm x 145mm x 385mm
Generator: 29.5 kg

X-ray Tube

Current Regulation
Load: <0.5% at max mA output over					the output kV range				
X-ray Tube Type					
Line: <0.5% at max mA output over		
X-ray Focal Spot Size			
the input line range load:
Ripple 						kV: <1.0% p-p of maximum output		mA: <1.0% p-p of maximum output

Beam Port		
			
			

Glass
0.8 mm as per IEC60336
Fan beam: 90º x 10º			(Beam port opening: 105º x 10º max)		
Cone beam : 40º

RepeatabilitykV: <0.5% 							mA: <0.5%

Operating Environment

Overshoot		 kV: <5% of rated output

Operating Temperature

0°C to 40°C

Output Rise Time 								Standard Rise Time ≤ 500 msec from
10%–90% of max rated output(Faster Rise Time available upon request)

Storage Temperature-40°C to 85°C

Thermal Cut Off		

60ºC ± 3ºC of oil temperature

Cooling			

Air Cooled

Humidity		 98% non-condensing

Radiation shield 		
Less than 0.5 mR/hr at 5cm from the 					surface of the chassis as per FDA 21 					CFR 1020.40

40

Generator

LED Indicators

446

145

40˚

POWER

Illuminated when Power is present

X-ray ON

Illuminated when Interlock is closed
and HV is enabled

ARC

ARC-ing fault

OCOver Current Fault

OT

Illuminated when oil temperature
exceeds 60±3ºC

OP

Illuminates when selected power
exceeds the rated power

OV

Over Voltage fault

384.5

65˚

105˚ 174.6±0.8
F.S.

247
61.4±0.8
F.S.

J1: AC Input
70

223±0.8
F.S.

Unit: mmControl Unit

N

NeutralGND

GroundL

230 VAC Input

J2 Connector: (Interlock 9 Pin Male)

254

247
70

Unit: mm

Pin Out

Name

1

Interlock out

2

Interlock In

3

X-ray On Relay contact Common

4

X-ray On Relay contact N/C

5

X-ray On Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

J3 Connector: (RS232 9 Pin Female)

Graphical User Interface

254

Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

41

Multipurpose X-Ray
Sources & Generators

5000

HVG Series - High Voltage Generators

4000

HVG 050

HVG450

HVG160

HVG225

160

225

1000

2000

Power (w)

3000

HVG3200

HVG10050100Voltage (KV)

320

450

HVG SeriesHigh Voltage Generators
The HVG Series of High Voltage Generators offers high stability
and reliability over a variety of voltage and current outputs.
Units can be unipolar or bi-polar and can be customized per
OEM requirements. The High Voltage Generators are compatible
with most commonly used X-ray tubes, allowing for effortless
OEM integration.
Features•••

Small Form Factor
Field Serviceable
Oil Insulation for ease of maintenanceApplications•
Industrial NDT•
Medical Imaging•
Security			HVG05050 kV, 5000 W
High Voltage Generator for 			Mammography
Applications
Digital mammography operations

Key Features• Compact and robust• Rapid output response time
• Modular design provides flexible mounting configurations• Filament with precision loop emission control
Specifications
Input Line Range

230 VAC, 50/60 Hz  Input Current

30A  

Output kV

10–50 kV

Output Power		5000 W peak output*6000 W peak output also available

Mammography Generator

Output mA

10 - 140 mA

Voltage Regulation

Line regulation: < 0.2% for Line Input      changes over specified range (with            constant ambient temperature and load).
Load regulation: < 1.0% for mA changes
over specified range (with constantambient temperature and line input).

Current Regulation
Line regulation: < 0.5% for Line Input                                               changes over specified range (with                                                     constant ambient temperature and load)kV regulation: < 0.5% for mA changes
over specified kV range (with constantambient temperature and line input)
Rise time≤5 msec  

Ripple 			≤1%

Repeatability		

<0.5%  for both KV and mA output

Output Connectors

High Voltage output: Claymount CA-3

Control Unit

Dimensions
Generator: 144mm x 125mm x 134mm
							
Control unit: 344mm x 254mm x 64mm
Weight			Generator: 4 Kg							Control unit: 4 Kg

Operating Environment
Operating Temperature

0°C to 40°C ·Storage temperature:-40°C to +85°C

Humidity:

90% non-condensing

Graphical User Interface

    

44

LED Indicators

Generator

Power
X-ray On

HIGH VOLTAGE OUTPUT
CLAYMOUNT CA3
CONNECTOR TO
CONTROL INVERTERHIGH VOLTAGE OUTPUT
CLAYMOUNT CA3
HIGH VOLTAGE OUTPUT
CLAYMOUNT CA3

125

CONTROL INVERTER

125 125

OCOver current fault

OT

Illuminated when oil temperature exceeds
60±3ºC.
Over Power fault
Over voltage fault

OP
OV

134

CONNECTOR TO
CONNECTORTOCONTROL
INVERTER

Arc

Illuminated when power is present
Illuminated when interlock
is closed and HV is enabled
ARC-ing fault

J1: AC Input
N
GND
L

134 134

144

NeutralGround230 VAC Input

J2: Analog and Interlock Control Interface
144
144

344

Unit: mmControl Unit

64
254
254

64

Pin Name
Out

Pin
Out

Name

1

Interlock out (+15V)

14

N/A

2

Interlock In

15

Detection enable3

Contactor common

16

N/A

4

X-ray OFF17

3.3Vref

5

X-ray ON

18

3.3Vref_Gnd

6

24VDC_Gnd

19

X-ray Enable

7

24VDC_OUT

20

N/A

8

EXT_kV Prog

21

External exposure

9

EXT_mA Prog

22

N/A

10

EXT_mA Gnd

23

Ready

11

EXT_kV Mon

24

Prep

12

EXT_mA Mon

25

Starter return

13

EXT_Mon Gnd

344
344

J3: RS232 Interface

344

64
254

64
254

Unit: mm

Pin
Out

Name

1

TX+

2

TX-

3
4

TX+
TX-

5
6

SIGNAL GND
N/A

7

N/A

8

N/A

9

N/A

45

HVG100
100 kV, 1000 W
High Voltage X-Ray Generators
Applications
Thickness Gauging, Tire Inspection, Industrial NDT, Food & Packaging Inspections

Key Features• High Frequency• Compact and Robust• Power Factor Corrected• User Friendly RS232 Digital Interface• Field Serviceable
Specifications
Input Line Range

230VAC, 50/60Hz, 10 Amps RMS

Output Current

0-25mA (Per X-ray tube ratings)

Output KV

30–100kV, negative or positive polarity

kV Rise Time		

< 1.0Sec from 10% to 90% of theoutput voltage

100 kV, 1 kW
Uni-polar Generator
Cooling			Self cooled, forced air

Dimensions 		

Generator: 254mm x 197mm x 252mm

Weight			Generator: 13kg		

kV Accuracy+/-2%

kV Stability< 25PPM per hour after 1 hourwarm-upkV Regulation≤ 0.01% of rated output voltage over
specified input voltage range≤ 0.01 % of rated output voltage over
specified mA range

Operating Environment

Output Ripple 		< 1% Peak – Peak at >20KHz               < 0.1% Peak – Peak at <20KHz

Operating Temperature

0°C to 40°C

Storage temperature:–40°C to +80°C

Humidity: 		

98% non-condensing

System Block Diagram (For Reference Only)

Output Power		1000 W maximum continuous output
			1500 W also available upon request
mA Rise Time

Available on special request

mA Regulation≤ 0.01% of rated output current over
specified input current range≤ 0.01 % of rated output current over
specified kV range

Safety and Regulatory
Compliances

Designed to meet IEC/EN 61010-1

Output Connectors
  Claymount CA1 HV connector (XR-7)
R10 (CA10) available upon
special request

Insulation		 Oil

ComputerUni-polar Generator

X-ray Tube

Cooler46

LED Indicators

Generator

OP

Over Power fault

OCOver current fault

Arc

ARC-ing fault

OT

Illuminated when oil temperature exceeds
60±3ºC.

OVOver voltage fault

X-ray On

Illuminated when interlock
is closed and HV is enabled

Power

Illuminated when power is present

J1: AC Input
N

NeutralGND

GroundL

230 VAC Input

J2: Interlock

Unit: mm

Graphical User Interface

Pin Out

Name

1

Interlock out

2

Interlock In

3

X-ray on relay contact common

4

X-ray on relay contact N/C

5

X-ray on Relay contact N/O

6

N/A

7

N/A

8

N/A

9

N/A

J3: RS232 Interface
Pin Out

Name

1

N/A

2

TX-

3

RX+

4

N/A

5

SIGNAL GRD

6

N/A

7

N/A

8

N/A

9

N/A

47

HVG160–450
160kV–450kV, 1.8KW–4.5KW

High Voltage Generator
Applications
Industrial NDT, Security Cargo Inspection, Medical Irradiation & Sterilization, General Purpose

Unipolar: HVG 160 & HVG 225
Key Features
•
•
•
•
•
•

Compact form factor
Oil based insulation for efficient thermal dissipation
Fully enclosed, fan-less design for use in rugged environments
Advanced HV design optimized for performance & reliability
Plug and play compatibility with major tube brands
Dual filament supplies controlled by closed-loop emission
current control

Model

HVG160CathodeAnode
Bipolar: HVG 320 & HVG 450

HVG225

HVG320

HVG450Max. Output Power

1.8KW

3.0KW

1.8KW

3.0KW

1.8KW

4.5KW

1.8KW

4.5KW

Output KV (see note)

10–160

10–160

10–225

10–225

10–320

10–320

10–450

10–450

Output mA (see note)

0–30

0–30

0–30

0–30

0–30

0–30

0–30

0–30

Output Polarity

-/+

-/+

-/+

-/+

Bipolar

Bipolar

Bipolar

Bipolar

R28

R28

kV: 0.05% peak-peak; mA: 0.1%

RippleStability

Less than 0.1% per 8hr after 1 hr warm-up

Repeatability

kV: ±40V; mA: 2uA

AccuracykV: ±1.0%; mA: ±0.5%

Output Rise Time
Output HV Connector
Operating Temperature
Storage Temperature
Humidity
Cooling
Duty Cycle
Dimension (mm)WeightInput Voltage
Input Current
Communication

Preset 3 sec (1–10 sec adjustable)
R24

12A

R24

R28

R28
R24
0˚C to +40˚C-40˚C to +80˚C
98% non-condensing
Forced Air Cool
100%

R24

280W x 560L x 448H

2x (280W x 560L x 448H)

60kg

120kg

20A

230VAC ±10%, 50/60Hz, Single Phase
12A
20A
12A
RS232 / Ethernet / Analog

30A

12A

30A

Note: Specific kV and mA range setting to be discussed for optimal performance.

48

Generator Dimensions

Unit: mmControl Interface Connections
Bipolar Module

Unipolar Module

CathodeAnode49

HVG160–450
160kV–450kV, 1.8KW–4.5KW
Graphical User Interface

LED Indicators
LED Indicators
X-RAY ON
OV

Illuminated when interlock Is closed and HV is enabledOver Voltage Fault

OC

Over Current Fault

OP

Over Power Fault When Exceeds Rated PowerPower
REG-ERROR
ARC
OT

Illuminated When Power Is Present
Regulation Error
Arcing Fault
Illuminated When Oil Temperature Exceeds 65±3ºC.

J2: Analog Interface (AMP747238-6)
Pin Out

Name

1

Interlock Out (15VDC)

Connectors

2Interlock In (15VDC)

Connectors

3

Relay Contactor (+24V)

4

LED (24VDC, 0.2 A Max))5

LED (24VDC Return)6

15V GND

7

15V Out

8

EXT_kV Program

9

EXT_mA Program

10

EXT_PROGRAM_GND

11

EXT_kV Monitor

12

EXT_mA Monitor

13

EXT_FIL_ENABLE

14

EXT_FIL_MONITOR

15

Sync In

16

X-ray Pre-warning

17

10V-REF (10mA Max)

Connector
J1
J2
J3
J4
J5
J6
J7

Description
Main & Auxiliary Input Power
Analog Interface
RS232 Digital Interface
RJ45 Ethernet Digital Interface
Anode VAC Input
Anode Feedback / Control
HV Connector

J1: Main & Auxiliary Input Power (MS3102A24-11P)
Pin Out

Name

A

AUX – 180-264VAC

B

AUX – GroundC
D

AUX – Neutral
Main – 180-264VAC

E
F

Main – Ground
Main – Neutral

18

10V-RET

19

Cooler Fault

G

N/A

H
I

N/A
N/A

20

FIL_Select21

External X-ray Enable

22

X-ray Pre-Warning Return

23

N/A

24

N/A

25

N/A

50

J3: RS232 Digital Interface

J6: Anode Feedback & Control (For Bipolar Configuration)

Pin Out

Name

Pin Out

Name

1

N/A

1

ANODE-kV-FDBK

2

TX-

2

N/A

3

RX+

3

N/A

4

N/A

4

ANODE-mA-FDBK

5Signal Ground

5

N/A

6

N/A

6

GND

7

N/A

7

ANODE-ARC-DETECT

8

N/A

8

N/A

N/A

9

ANODE-INV-DRIVE A

10ANODE-INV-DRIVE B

11

ANODE INV-SENSE

12ANODE-INV-TEMP-SENSEANODE OIL TEMP

9

J4: RJ45 Ethernet Digital Interface
Pin Out

Name

1

TX+

13

2

TX-

14

+15V OUT

3

RX+

15

-15V OUT

4

N/A

5

N/A

6

RX-

7

Ground8

GroundJ5: Anode VAC Input (For Bipolar Configuration)
Pin Out

Name

A

GND

B

NEUTRAL

C

LINE

System Interconnection

J7: High Voltage Connector (R28 or R24)
Pin OutCSLName
HV Output
Small Filament Output
Large Filament Output

Accessories and Options
Accessories and Options

Accessories and options offer convenience and ease of adaptation
with IXS and HVG products.
•

Collimators•

Coolers•

External Battery and Convertor•

Options for Portable Source IXS1212•

Recommended Kit for Electronic Protection•

Reusable Cases

Collimators•••

Generator shown with
fan-shaped lead Collimator

Provides convenience for mechanical interface between IXS and the system
Minimizes radiation leakage
Ability to shape x-ray beam

DF3000-009

Asymmetrical Fan Beam 80˚, 160kV with Flange

DF3000-063

Cone Beam 40˚, 160kV

DF3000-142

Cone Beam 40˚, 160kV with Flange

DF3000-096

Symmetrical Fan Beam 75˚, 160kV

DF3000-111

Symmetrical Fan Beam 56˚, 100kV Mid-Size

DF3000-082

Symmetrical Fan Beam 60˚, 80kV Compact

Generator shown withconical lead Collimator

Cooler•••Ability to operate IXS sources in high temperature environment
Enhanced stability
Allows sources to run continuous duty cycles

Model: AS3001-126

An IXS1650 x-ray source
configured with a cooler(For illustration purposes)

53

Accessories and Options

External Batteries and Converters••••

36VDC Battery pack, AC converter and charger
Supports 500W IXS source operations for 2.5 hours
Charging time: 8 hours
Ideal for mobile/field uses

The x-ray source, the detector, and the computer ofthis mobile system are powered by the battery

Options for Portable IXS1212

Laser Alignment Guide

Battery pack & charging station

Generator shown with protective
cover and handle

AS3001-288M

Battery pack( inc. 24V LiFeO4 battery)

AS3001-300M

Charging Station

DS3000-041M

Protective Cover and Handle

DB3000-122M

Tripod Mount

DS3000-106

Laser Alignment Guide

JP3000-019

Carrying Case

Generator shown mounted
on tripod

54
Wireless option

Recommended Kit for Electronic Protection••

EMI Filter

Protects electronics from external factors, such as input power
instability, transient surge, or electromagnetic disruption
Recommend every system to equip with adequate isolationtransformer and EM1 filter, Consult with VJX sales forrecommended kit.
EC3000-158

Filter, EMI

ET3000-122
ET3000-123

Isolation Transformer, for 100W
Isolation Transformer, for 150W

ET3000-124

Isolation Transformer, for 200W

ET3000-125

Isolation Transformer, for 500W

Isolation Transformer

Reusable Cases••

Rugged and reusable
Ideal for field use and repair center for product transport

JP3000-005

Case for IXS 160kV

JP3000-018
JP3000-019

Case for IXS 100kV
Case for IXS 120kV Portable

55

Founded in 1987, VJ Technologies, Inc. is a world leader in thedesign, manufacture, installation, operation and long-term
support of a wide range of industrial x-ray systems. Thesesystems utilize state of the art digital imaging. In almost three
decades, we’ve grown in size around the world and expanded
the range of industries we serve and products offered.
Our customer base includes companies in the Aerospace,
Automotive, Defense, Electronics, Homeland Security, Nuclear,and Oil & Gas fields. The design engineering and manufacturingexpertise we have accumulated thus far drives our systems,
products and services to become must-haves in the X-ray
inspection world.
Our mission is to meet the unique X-ray inspection needs ofeach client that calls upon our experience and expertise. We
accomplish this by designing, manufacturing, and supportinga variety of unique and customized inspection system needs,
which often include our world-renowned, innovative line of
X-ray sources & generators.

Since 2008, the components business unit, VJ X-ray, has
become our specialization and central to our growth. We havecommitted ourselves to develop and maintain a state-of-the-art
line of Integrated X-ray Sources and High Voltage Generators.
Our rapidly growing OEM business has expanded our markets
into Security, Food & Pharmaceutical, and Medical Imaging.
We have an install base of more than 10,000 units since 2010.
The VJ professional service engineers perform high quality inhouse inspection services for those customers with low-volume
or “one-off” inspection needs when they are not out in the field
supporting 1000+ installed systems worldwide.
Our global presence is headquartered on Long Island, 50 miles
east of New York City in Bohemia, New York. We have offices
and facilities throughout the US, Asia, and Europe.

89 Carlough Road, Bohemia, NY 11716, USA
Email: info@vjt.com, Tel: +1 631 589 8800 ext. 1103				
www.vjtxray.com 																														Rev. 3, September 2016