GP01198C/33/JA/01.22
70209134

Products

有効なバージョン

Solutions

V 01.04 以上（機器ファームウェア）

機能説明書
J22 TDLAS ガスアナライザ
Modbus TCP および RS485

Services機能説明書2

J22 TDLAS ガスアナライザ

Endress+Hauser

J22 TDLAS ガスアナライザ

3.7 Diagnostics ................................................................81

目次
1

機能説明書

本説明書について ............................................... 4

1.1 警告 ............................................................................ 4
1.2 機器のシンボル........................................................... 4
1.3 米国輸出管理規則の遵守 ............................................ 4
1.4 資料の機能 .................................................................. 5
1.4.1

対象グループ ......................................................... 5

1.5 本書の使用法 .............................................................. 5
1.5.1

本書の構成 ............................................................. 5

1.5.2

パラメータ説明の構成 ......................................... 6

1.6 使用されるシンボル ................................................... 6
1.6.1

各種情報を示すシンボル ..................................... 6

1.6.2

図中のシンボル ..................................................... 7

1.7 関連資料 ..................................................................... 7
1.7.1

4

Expert メニューの概要 ....................................... 8

3

機器パラメータの説明 ........................................ 10
Display ...................................................................12

3.1.2

Configuration backup ..........................................22

3.1.3

Diagnostic handling.............................................24

3.1.4

Administration .....................................................26

3.2 Sensor ....................................................................... 31

3.7.2

Event logbook .......................................................86

3.7.3

Device information ..............................................87

3.7.4

Main electronic module + I/O module 1...........89

3.7.5

Sensor electronic module (ISEM) ......................90

3.7.6

I/O module 2.........................................................91

3.7.7

I/O module 3.........................................................92

3.7.8

Display module .....................................................93

3.7.9

Data logging..........................................................93

3.7.10Heartbeat Technology .........................................97

3.7.11

Simulation .......................................................... 110

3.7.12

Spectrum plots................................................... 114

3.7.13

SD card ................................................................ 119

認定固有の初期設定 ....................................... 121
4.1.1

システム単位 .................................................... 121

4.1.2

フルスケール値 ................................................ 121

4.1.3

電流出力スパン ................................................ 121

4.2 US 単位 ................................................................... 121

3.1 System ...................................................................... 11
3.1.1

Diagnostic list .......................................................83

4.1 SI 単位 .................................................................... 121

標準資料.................................................................. 7

2

3.7.1

5

4.2.1

システム単位 .................................................... 121

4.2.2

フルスケール値 ................................................ 121

4.2.3

電流出力スパン ................................................ 121

単位の短縮表記の説明 .................................... 122

5.1 SI 単位 .................................................................... 122
5.2 US 単位 ................................................................... 122

3.2.1

Measured values ..................................................31

3.2.2

System units .........................................................38

3.2.3

Stream ...................................................................423.2.4

Dew point ..............................................................43

3.2.5

Peak tracking ........................................................46

6.1.1

レジスタ情報の構造 ........................................ 123

3.2.6

Sensor adjustment ...............................................47

6.1.2

アドレスモデル ................................................ 123

3.2.7

Stream change compensation ............................49

3.2.8

Calibration.............................................................51

3.3 I/O configuration ....................................................... 52
3.4 Input ......................................................................... 54
3.4.1

Current input 1～n ..............................................54

3.5 Output ....................................................................... 57
3.5.1

Current output 1～n ............................................57

3.5.2

Switch output 1 ....................................................62

3.5.3

Relay output 1～n ................................................67

3.6 Communication ......................................................... 71
3.6.1

Modbus configuration .........................................71

3.6.2

Modbus information............................................76

3.6.3

Modbus data map ................................................77

3.6.4

Web server ............................................................77

Endress+Hauser

5.3 英国単位 ................................................................. 122

6

Modbus レジスタ情報 .................................... 123

6.1 注記 ........................................................................ 123

6.2 エキスパート操作メニューの概要.......................... 124
6.3 レジスタ情報 .......................................................... 133
6.3.1

System サブメニュー....................................... 133

6.3.2

Sensor ................................................................. 137

6.3.3

I/O configuration サブメニュー..................... 143

6.3.4

Input サブメニュー .......................................... 144

6.3.5

Output サブメニュー ....................................... 145

6.3.6

Communication サブメニュー ....................... 147

6.3.7

Diagnostics......................................................... 150

6.3.8

Simulation .......................................................... 158

6.3.9

Spectrum plots................................................... 158

6.3.10SD card ................................................................ 159

3

J22 TDLAS ガスアナライザ

機能説明書

1

本説明書について

1.1 警告情報の構造

意味
危険な状況を警告するシンボルです。この状況を回避できなかった場

原因（/結果）合、重傷または致命傷を負う可能性があります。

違反した場合の結果（該当する場合）‣ 是正処置危険な状況を警告するシンボルです。この状況を回避できなかった場
原因（/結果）合、軽傷または中程度の傷害を負う可能性があります。

違反した場合の結果（該当する場合）‣ 是正処置器物を損傷する可能性がある状況を警告するシンボルです。
原因/状況
違反した場合の結果（該当する場合）‣ アクション/注記
表 1. 警告

1.2 機器のシンボル
シンボル

説明
レーザー放射シンボルは、J22 TDLAS ガスアナライザの使用時に危険な可視レーザー光に暴露する危険性をユーザーに警告するものです。
高電圧シンボルは、人体に危害を与えるほどの高電位の存在を作業員に警告するものです。一部の産業では、特定のしきい値を超える高電圧を指します。高電圧のかかる機器や導体については、特別な安全要件と安全手順を満たす必要があります。
ETL マークは、製品が北米安全基準に適合していることを証明するものです。米国とカナダの規制監
督機関（AHJ）および規制担当官は、製品が公示されている産業基準に適合する証明として ETL マー
クを承認します。
WEEE シンボルは、本製品を未分別の廃棄物として廃棄することが禁止されており、回収/再利用のために分別回収施設に送る必要があることを示します。
CE マーキングは、欧州経済領域（EEA）内で販売される製品について、健康、安全、環境に関する保護基準に適合していることを示します。
表 2. シンボル

1.3 米国輸出管理規則の遵守
Endress+Hauser は、そのポリシーにおいて、米国商務省のウェブサイト Bureau of Industry and Security で詳述されている米国輸出管理法を厳格に遵守しています。

4Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

1.4 資料の機能
本資料は取扱説明書の一部であり、パラメータの参照資料として、操作メニューの各パラメータに関する詳細説明が記載されています。

1.4.1

対象グループ

本資料は、全ライフサイクルにわたって本機器を使用し、特定の設定を行うスペシャリストのために用意されたものです。本書は機器の機能に関してより詳細な知識が要求される作業を実行する場合に使用します。
••••

各種条件下における測定の設定
各種条件下における測定の最適化
通信インターフェースの詳細設定
難しいケースにおけるエラー診断

1.5 本書の使用法
1.5.1

本書の構成

本書には Expert メニュー → の構成に応じたサブメニューとそのパラメータが記載されており、これは、ユーザーの役割「メンテナンス」が有効になった場合に表示されます。

A0029160-EN

1

Endress+Hauser

操作メニュー配置のサンプル図5

機能説明書

J22 TDLAS ガスアナライザ

 Operation メニュー、Setup メニュー、Diagnostics メニューのメニュー構造に応じたパラメータ配置に関する
追加情報と簡単な説明については、取扱説明書 → を参照してください。
 操作メニューの操作コンセプトについても、取扱説明書 → を参照してください。

1.5.2

パラメータ説明の構成

パラメータ説明は次のセクションに個別に記載されています。
完全なパラメータ名

説明
現場表示器またはウェブブラウザを使用する場合のパラメータのナビゲー

ナビゲーション

ションパス
操作ツールを使用する場合のパラメータのナビゲーションパス
メニュー、サブメニュー、パラメータの名前は、表示器や操作ツールに表示される形式に応じて短縮されます。
必須条件

このパラメータは、特定の条件下でのみ使用できます。

説明

パラメータ機能の説明

選択項目

パラメータの個々のオプションのリスト▪ オプション 1▪ オプション 2

ユーザー入力パラメータの入力範囲

ユーザーインターフェース

パラメータの表示値/データ

初期設定

工場出荷時の初期設定

追加情報

以下のような追加説明▪ 個別オプション関連▪ 表示値/データ関連▪ 入力範囲関連▪ 初期設定関連▪ パラメータ機能関連

1.6 使用されるシンボル
1.6.1

各種情報を示すシンボル

シンボル

説明
ヒント

A0011193

追加情報を示します。
資料参照

A0028658

6

Endress+Hauser

J22 TDLAS ガスアナライザ

シンボル

機能説明書

説明
ページ参照

A0028659

図参照
A0028660

現場表示器による操作
A0028662

操作ツールによる操作
A0028663

アクセスコードで保護されているパラメータ
A0028665

1.6.2

図中のシンボルシンボル

説明

1, 2, 3 ...

項目番号

A、B、C ...

図A-A、B-B、C-C ...

断面図

1.7 関連資料
1.7.1

標準資料

資料番号

資料の種類

説明

BA02152C

取扱説明書

機器の設置、設定、メンテナンスに必要なすべての操作や手順を網羅した概要書です。

XA02708C

安全上の注意事項

作業員や機器の安全性を確保するための J22 TDLAS ガスアナライザの設
置や操作における必要条件が記載されています。

XA03086C

安全上の注意事項

作業員や機器の安全性を確保するための J22 TDLAS ガスアナライザの設INMETRO

置や操作における必要条件が記載されています。INMETRO 認証用の資料です。

XA03087C

安全上の注意事項

作業員や機器の安全性を確保するための J22 TDLAS ガスアナライザの設

JPNEx

置や操作における必要条件が記載されています。JPNEx 認証用の資料です。

XA03090C

安全上の注意事項

作業員や機器の安全性を確保するための J22 TDLAS ガスアナライザの設PESO/KC

置や操作における必要条件が記載されています。PESO/KC 認証用の資料です。

TI01607C

技術仕様書

機器の計画支援。
本書には、アナライザに関するすべての技術データが記載されています。

Endress+Hauser

7

機能説明書2

J22 TDLAS ガスアナライザ

Expert メニューの概要

以下の表は、エキスパート操作メニューとそのパラメータのメニュー構造の概要を示したものです。サブメニュー
またはパラメータの説明については、参照ページをご覧ください。
ExpertLocking status→

10

User role→

11

Enter access code→11

‣ System→

12‣ Display→11‣ Configuration backup→

22‣ Diagnostic handling→24‣ Administration

→

26

→

31

‣ Measured values→

31‣ System units→

39‣ Stream→42‣ Dew point→

43‣ Peak tracking→

44‣ Sensor adjustment→

47‣ Stream change
compensation (SCC)→49‣ Calibration→

51‣ I/O configuration→52‣ Input

→

54

→

54

→

57

‣ Current output 1～n→

57‣ Switch output 1～n→62‣ Sensor‣ Current input 1～n‣ Output

8

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書‣ Relay output 1～n‣ Communication‣ Modbus configuration‣ Modbus information‣ Modbus data map‣ Web server‣ Diagnostics
Actual diagnostics
Previous diagnostics
Operating time from restart
Operating time
‣ Diagnostic list
‣ Event logbook‣ Device information‣ Main electronic module +I/O module 1‣ Sensor electronic module(ISEM)‣ I/O module 2‣ I/O module 3‣ Display module‣ Data logging‣ Heartbeat Technology‣ Simulation‣ Spectrum plots‣ SD card

Endress+Hauser

→

67

→

71

→

71

→

76

→

77

→

77

→

81

→

81

→

82

→

82

→

82

→

83

→

86

→

87

→

89

→

90

→

91

→

92

→

93

→

93

→

97

→

110

→

115

→

119

9

機能説明書3

J22 TDLAS ガスアナライザ

機器パラメータの説明

以下のセクションでは、現場表示器のメニュー構造に従ってパラメータが記載されています。操作ツール用の特定のパラメータは、メニュー構造の該当する箇所に示されます。
ExpertLocking status→

10

User role→

11

Enter access code→11

‣ System→

12‣ Sensor→

31‣ I/O configuration→52‣ Input→54‣ Output→

57‣ Communication→

71‣ Diagnostics→

81

Locking status
ナビゲーション
説明

Expert → Locking status
有効な書き込み保護設定を表示します。

ユーザーインター ▪ Hardware locked
フェース▪ Temporarily locked

追加情報

ユーザーインターフェース
2 種類以上の書き込み保護設定が有効な場合は、最も優先度の高い書き込み保護設定が現場表示器に示されます。操作ツールの方は、有効な全種類の書き込み保護設定が表示されます。
 アクセス権の詳細については、本機器の取扱説明書 →の「ユーザーの役割と関連するアクセス権」および「操作コンセプト」セクションを参照してください。

選択項目
選択項目

説明

None

Locking status パラメータ →に表示されるアクセス権が適用されます。現場表示器にのみ表示されます。
Hardware locked

ハードウェア書き込みロック用の DIP スイッチが PCB 基板で有効になっ（優先度 1）ています。これにより、パラメータへの書き込みアクセスがロックされます（例：現場表示器または操作ツールを介して）。

10

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

選択項目

説明

Temporarily

機器の内部処理（例：データアップロード/ダウンロード、リセットな

lockedど）を実行中のため、パラメータへの書き込みアクセスが一時的にロッ（優先度 4）

クされます。内部処理が完了すると、再びパラメータを変更することが可能です。

User role
ナビゲーション
説明

Expert → User role
現場表示器、ウェブブラウザまたは操作ツールを介したパラメータへのアクセス権を表示します。

ユーザーインター Operator
フェース

Maintenance

初期設定

Maintenance

追加情報

アクセス権は、Enter access code パラメータ →から変更できます。

また、書き込み保護機能が有効な場合は、それによって現在のアクセス権がさらに制限されます。
 アクセス権の詳細については、本機器の取扱説明書 → の「ユーザーの役割と関連するアクセス権」および「操作コンセプト」セクションを参照してください。

Enter access code
ナビゲーション
説明

Expert → Ent. access code
この機能を使用して、パラメータ書き込み保護を解除するためのユーザー固有のリリー
スコードを入力します。

ユーザー入力数字、英字、特殊文字から成る最大 16 桁の文字列

初期設定

0000; ユーザーが変更可能

追加情報

ログインの手順や説明については、J22 取扱説明書 BA02152C →を参照してください。

3.1 System
ナビゲーション

Expert → System‣ SystemEndress+Hauser→

12‣ Display→11‣ Configuration backup→

22‣ Diagnostic handling→24‣ Administration

→

26

11

機能説明書3.1.1

J22 TDLAS ガスアナライザ

Display

ナビゲーション

Expert → System → Display‣ Display

12

Display language→

13

Format display→

13

Value 1 display→

140% bargraph 1→

14100% bargraph 1→

14

Decimal places 1→

15

Value 2 display→

16

Decimal places 2→

16

Value 3 display→

170% bargraph 3→

17100% bargraph 3→

17

Decimal places 3→

17Value 4 display→

17

Decimal places 4→18

Display interval→

18

Display damping→

19

Header→20

Header text→

20

Separator→

21

Contrast display→

21

Backlight→

21

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Display language
Expert → System → Display→ Display language

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、設定された言語を現場表示器で選択します。

選択項目

EnglishFrançais
Italianoрусский язык (Russian)
中文 (Chinese)

初期設定

English（または、ご注文の言語が機器にプリセットされます）

Format display
Expert → System → Display → Format display

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、現場表示器に表示する測定値の形式を選択します。

選択項目

1 value, max. size
1 bargraph + 1 value
2 values
1 value large + 2 values
4 values

初期設定

1 value, max. size

追加情報

説明
表示形式（サイズ、バーグラフなど）および同時に表示する測定値の数（1～4）を設定できます。この設定は通常の測定にのみ有効です。
Value 1 display パラメータ →～Value 4 display パラメータを使用して、現場表示器に表示

する測定値と表示順序を指定します。
選択した表示モードで許容される数より多くの測定値を指定した場合は、機器表示部上で値が交互に表示されます。次の表示に切り替わるまでの表示時間は、Display interval パラメー
タ→で設定します。

現場表示器に表示できる測定値：「1 value, max. size」オプション
H2O Analyzer

46.21
ppmv
A0016529

Endress+Hauser

13機能説明書

J22 TDLAS ガスアナライザ「1 bargraph + 1 value」オプション

H2O Analyzer

46.21 ppmv

89.43
°F
A0013098「2 values」オプション

H2O Analyzer

46.21
ppmv

89.43
°F
A0013100「1 value large + 2 values」オプション
H2O Analyzer

46.21
ppmv

89.43 °F
P

0.97 bar
A0013102「4 values」オプション
H2O Analyzer
46.21 ppmv
61.91 °C
P

0.97 bar-2.02 °C
A0013103

Value 1 display
ナビゲーション

Expert → System → Display→ Value 1 display

必須条件

現場表示器があること。

説明

この機能を使用して、現場表示器に表示する測定値の 1 つを選択します。

14

Endress+Hauser

J22 TDLAS ガスアナライザ

選択項目

機能説明書

▪ Concentration▪ Dew point 1▪ Dew point 2

1
0F

1

▪ Cell gas pressure▪ Cell gas temperature
初期設定

Concentration

追加情報

説明
複数の測定値が同時に表示される場合は、ここで選択した測定値が最初の値として表示されます。この値は通常の測定時にのみ表示されます。
Format display パラメータ →を使用して、同時に表示する測定値の数と表示形式を設定

します。

依存関係
表示する測定値の単位として、System units →の設定が使用されます。

0% bargraph value 1
Expert → System → Display → 0% bargraph 1

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、測定値 1 のバーグラフ表示の 0% の値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

ppmv

追加情報

説明
Format display パラメータ →を使用して、バーグラフとして表示する測定値を設定します。

ユーザー入力
表示する測定値の単位として、System units →の設定が使用されます。

100% bargraph value 1
Expert → System → Display → 100% bargraph 1

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、測定値 1 のバーグラフ表示の 100% の値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

ppmv

追加情報

説明
Format display パラメータ →1を使用して、バーグラフとして表示する測定値を設定します。

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser

15

機能説明書

J22 TDLAS ガスアナライザ

ユーザー入力
表示する測定値の単位として、System units →の設定が使用されます。

Decimal places 1
Expert → System → Display → Decimal places 1

ナビゲーション
必須条件

Value 1 display パラメータ →で、測定値が設定されていること。

説明

この機能を使用して、測定値 1 の小数点以下の桁数を選択します。

選択項目

符号付き浮動小数点数
▪ x
▪ x.x
▪ x.xx▪ x.xxx▪ x.xxxx

初期設定

x.xx

追加情報

説明
この設定は、機器の測定や値計算の精度には影響しません。

Value 2 display
Expert → System → Display → Value 2 display

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、現場表示器に表示する測定値を選択します。

ユーザー入力選択リストについては、Value 1 display パラメータ →

初期設定

None

追加情報

説明

を参照してください。

複数の測定値が同時に表示される場合は、ここで選択した測定値が 2 つ目の値として表示されます。この値は通常の測定時にのみ表示されます。
Format display パラメータ →を使用して、同時に表示する測定値の数と表示形式を設定

します。

依存関係
表示する測定値の単位として、System units →の設定が使用されます。

Decimal places 2Expert → System → Display → Decimal places 2

ナビゲーション
必須条件

Value 2 display パラメータ →で、測定値が設定されていること。

説明

この機能を使用して、測定値 2 の小数点以下の桁数を選択します。

選択項目

▪ x
▪ x.x
▪ x.xx16

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書▪ x.xxx▪ x.xxxx
初期設定

x.xx

追加情報

説明
この設定は、機器の測定や値計算の精度には影響しません。

Value 3 display
Expert → System → Display → Value 3 display

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、現場表示器に表示する測定値を選択します。

選択項目

選択リストについては、Value 1 display パラメータ →

初期設定

None

追加情報

説明

を参照してください。

複数の測定値が同時に表示される場合は、ここで選択した測定値が 3 つ目の値として表示されます。この値は通常の測定時にのみ表示されます。
Format display パラメータ →を使用して、同時に表示する測定値の数と表示形式を設定

します。

選択項目
表示する測定値の単位として、System units →の設定が使用されます。

0% bargraph value 3
Expert → System → Display → 0% bargraph 3

ナビゲーション
必須条件

Value 3 display パラメータ →説明

この機能を使用して、測定値 3 のバーグラフ表示の 0% の値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

None

追加情報

説明
Format display パラメータ →で、項目が選択されていること。

を使用して、バーグラフとして表示する測定値を設定します。

ユーザー入力
表示する測定値の単位として、System units →の設定が使用されます。

100% bargraph value 3
ナビゲーション

Expert → System → Display → 100% bargraph 3

必須条件

Value 3 display パラメータ →説明

この機能を使用して、測定値 3 のバーグラフ表示の 100% の値を入力します。

ユーザー入力

符号付き浮動小数点数

Endress+Hauser

で、項目が選択されていること。

17

機能説明書

J22 TDLAS ガスアナライザ

初期設定

None

追加情報

説明
Format display パラメータ →を使用して、バーグラフとして表示する測定値を設定します。

ユーザー入力
表示する測定値の単位として、System units →の設定が使用されます。

Decimal places 3Expert → System → Display → Decimal places 3

ナビゲーション
必須条件

Value 3 display パラメータ →で、測定値が設定されていること。

説明

この機能を使用して、測定値 3 の小数点以下の桁数を選択します。

選択項目

▪ x
▪ x.x
▪ x.xx▪ x.xxx▪ x.xxxx

初期設定

x.xx

追加情報

説明
この設定は、機器の測定や値計算の精度には影響しません。

Value 4 display
Expert → System → Display → Value 4 display

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、現場表示器に表示する測定値を選択します。

選択項目

選択リストについては、Value 1 display パラメータ →

初期設定

None

追加情報

説明

を参照してください。

複数の測定値が同時に表示される場合は、ここで選択した測定値が 4 つ目の値として表示されます。この値は通常の測定時にのみ表示されます。
Format display パラメータ →を使用して、同時に表示する測定値の数と表示形式を設定

します。

選択項目
表示する測定値の単位として、System units →の設定が使用されます。

Decimal places 4
ナビゲーション

Expert → System → Display → Decimal places 4

必須条件

Value 4 display パラメータ →説明

この機能を使用して、測定値 4 の小数点以下の桁数を選択します。

18

で、測定値が設定されていること。

Endress+Hauser

J22 TDLAS ガスアナライザ

選択項目

機能説明書

▪ x
▪ x.x
▪ x.xx▪ x.xxx▪ x.xxxx

初期設定

x.xx

追加情報

説明
この設定は、機器の測定や値計算の精度には影響しません。

Display interval
Expert → System → Display → Display interval

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、表示部上に交互に表示する測定値の表示時間の長さを入力します。

ユーザー入力1～10 s

初期設定

5s

追加情報

説明
選択された表示形式で同時に表示可能な数を、設定された測定値の数が超えた場合に限り、自動的にこの交互表示タイプとなります。
▪ Value 1 display パラメータ →～Value 4 display パラメータ →で、現場表示器に表示

する測定値を指定します。
▪ 測定値の表示形式は、Format display パラメータ →で設定します。

Display damping
Expert → System → Display → Display damping

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、プロセス条件に起因する測定値の変動に対する、現場表示器の応答時間の時定数を入力します。

ユーザー入力0.0～999.9 s

初期設定

0.0 s

追加情報

ユーザー入力1

この機能を使用して、表示のダンピングの時定数を入力します（PT1 エレメント ）。
1F▪ 小さな時定数を入力した場合、表示部は変動する測定変数に対して非常に素早く反応します。
▪ 一方、大きな時定数を入力した場合は、表示の応答性が遅くなります。
▪ 0 を入力するとダンピングはオフになります（初期設定）。

1

1 次遅延を伴う比例伝送特性

Endress+Hauser

19

機能説明書

J22 TDLAS ガスアナライザ

HeaderExpert → System → Display → Header

ナビゲーション
必須条件

現場表示器があること。

説明

この機能を使用して、現場表示器のヘッダーの内容を選択します。

選択項目▪ Device tag▪ Free text

初期設定

Device tag

追加情報

説明
ヘッダーテキストは通常の測定時にのみ表示されます。

A0029422

1 表示部のヘッダーテキストの位置
選択項目

20

Device tag は、Device tag パラメータ →で設定します。

Free text は、Header text パラメータ →で設定します。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Header text
Expert → System → Display → Header text

ナビゲーション
必須条件

Header パラメータ →で、Free text が選択されていること。

説明

この機能を使用して、現場表示器のヘッダー用にユーザー固有のテキストを入力します。

ユーザー入力最大 12 文字（英字、数字、特殊文字（例：@、%、/）など）

初期設定

.（点）

追加情報

説明
ヘッダーテキストは通常の測定時にのみ表示されます。

A0029422

1 表示部のヘッダーテキストの位置
ユーザー入力表示される文字数は使用される文字に応じて異なります。

Separator
ナビゲーション

Expert → System → Display → Separator

必須条件

現場表示器があること。

説明

この機能を使用して、桁区切り記号を選択します。

選択項目▪ .（点）▪ ,（コンマ）

初期設定------------

Contrast display
ナビゲーション

Expert → System → Display → Contrast display

必須条件

現場表示器があること。

説明

この機能を使用して、周囲条件（照明、読み取り角度など）に合わせて表示部のコントラス
トを調整するための値を入力します。

ユーザー入力20～80 %

初期設定

デフォルト値：50 %Backlight
ナビゲーション
必須条件
Endress+Hauser

Expert → System → Display → Backlight
現場表示器があること。
21

機能説明書

J22 TDLAS ガスアナライザ

説明

この機能を使用して、現場表示器のバックライトをオン/オフします。

選択項目▪ Disable▪ Enable
Enable

初期設定

3.1.2

Configuration backup

ナビゲーション

Expert → System → Configuration Backup‣ Configuration backup→

22

Operating time→

22

Last backup→

22

Configuration management→

22

Backup state→

23

Comparison result→

23

Operating Time
Expert → System → Configuration backup → Operating time

ナビゲーション

この機能を使用して、機器の稼働時間を表示します。

説明

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）

ユーザーインターフェース

追加情報

最大日数は 9999 です。これは、27 年に相当します。

Last backup
Expert → System → Configuration backup → Last backup

ナビゲーション

データのバックアップコピーが最後に機器メモリに保存されてからの時間を表示し説明ます。
ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）

Configuration management
ナビゲーション

Expert → System → Configuration backup → Configuration management

説明

この機能を使用して、機器メモリにデータを保存するための操作を選択します。

選択項目▪ Cancel▪ Execute backup

22

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書▪ Restore

1

2F▪ Clear backup data1

▪ Compare
初期設定

Cancel追加情報

選択項目▪ Cancel：何も実行せずにこのパラメータを終了します。
▪ Execute backup：現在の機器設定のバックアップコピーを HistoROM バックアップから機器のメモリに保存します。バックアップコピーには機器の変換器データが含まれます。現
場表示器に以下のメッセージが表示されます。Backup active, please wait!
1▪ Restore ：機器設定の最後のバックアップコピーを機器メモリから機器の HistoROM バッ
クアップに復元します。バックアップコピーには機器の変換器データが含まれます。現場
表示器に以下のメッセージが表示されます。Restore active! Do not interrupt power supply!
▪ Clear backup data：機器のメモリから機器設定のバックアップコピーを削除します。現場
表示器に以下のメッセージが表示されます。Deleting file1

▪ Compare ：機器メモリに保存された機器設定と HistoROM バックアップの現在の機器設定を比較します。現場表示器に以下のメッセージが表示されます。Comparing files
結果は Compar. result パラメータに表示されます。

HistoROM
HistoROM は、EEPROM タイプの不揮発性メモリです。

Backup state
ナビゲーション

Expert → System → Configuration backup → Backup state

説明

データバックアップ処理のステータスが表示されます。

ユーザーインター▪ Noneフェース▪ Backup in progress▪ Restoring in progress▪ Delete in progress▪ Compare in progress▪ Restoring failed▪ Backup failed

初期設定

None

Comparison result
ナビゲーション

Expert → System → Configuration backup → Compar. result

説明

機器メモリと HistoROM のデータ記録に関する最後の比較結果を表示します。

ユーザーインター▪ Settings identicalフェース▪ Settings not identical▪ No backup available▪ Backup settings corrupt▪ Check not done1

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser

23

機能説明書

J22 TDLAS ガスアナライザ▪ Dataset incompatible
初期設定

Check not done

追加情報

説明
Configuration management パラメータ →で Compare を選択すると、比較が開始されます。

選択項目▪ Settings identical：HistoROM の現在の機器設定と機器メモリのバックアップコピーは一致します。
Configuration management パラメータで、HistoROM を使用して他の機器の変換器設定
を機器に伝送した場合、HistoROM の現在の機器設定は機器メモリのバックアップコピーと一部しか一致しません。変換器の設定は一致しません。
▪ Settings not identical：HistoROM の現在の機器設定と機器メモリのバックアップコピー
は一致しません。
▪ No backup available：HistoROM の機器設定のバックアップコピーが機器メモリにはありません。
▪ Backup settings corrupt：HistoROM の現在の機器設定が破損しているか、または機器メ
モリのバックアップコピーとの互換性がありません。
▪ Check not done：HistoROM の機器設定と機器メモリのバックアップコピーとの比較がまだ完了していません。
▪ Dataset incompatible：機器メモリのバックアップコピーは機器と互換性がありません。

HistoROM
HistoROM は、EEPROM タイプの不揮発性メモリです。

3.1.3

Diagnostic handling

ナビゲーション

Expert → System → Diagnostic Handling‣ Diagnostic handling
Alarm delay→24‣ Diagnostic behavior→

25

Alarm delay
Expert → System → Diagnostic handling → Alarm delay

ナビゲーション
説明

この機能を使用して、機器が診断メッセージを生成するまでの時間間隔を入力します。
診断メッセージは遅延時間なしでリセットされます。

ユーザー入力0～60 s

初期設定

0s

追加情報

結果
この設定は、以下の診断メッセージに影響を及ぼします。
▪ 832 Electronics temperature too high▪ 833 Electronics temperature too low▪ 904 Cell gas flow not detected

24

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Diagnostic behavior サブメニュー
診断情報の各項目には、工場出荷時に特定の診断動作が割り当てられています。特定の診断情報については、ユーザーがこの割当てを Diagnostic behavior サブメニューで変更できます。
Diagnostic no. xxx パラメータでは、以下の項目を選択できます。
Alarm

機器が測定を停止します。Modbus RS485 を使用した測定値出力が、設定されたアラーム状
態になります。診断メッセージが生成されます。
バックライトが赤に変わります。

Warning

機器は測定を継続します。Modbus RS485 を使用した測定値出力は影響を受けません。診断
メッセージが生成されます。

Logbook entry only 機器は測定を継続します。診断メッセージは、Event logbook サブメニュー →され、操作画面表示と交互に表示されることはありません。
Off

にのみ表示

診断イベントは無視され、診断メッセージの生成または入力は行なわれません。

 すべての診断イベントの一覧については、機器の取扱説明書 →ナビゲーション

を参照してください。

Expert → System → Diagnostic handling → Diagnostic Behavior‣ Diagnostic behavior
Diagnostic no. 302→

25

Diagnostic no. 441→

25

Diagnostic no. 444→

25

Diagnostic no. 905→

26

Diagnostic no. 302（Device verification active）
ナビゲーション

Expert → System → Diagnostic handling → Diagnostic behavior → Diagnostic no. 302

説明診断メッセージ 302 Device verification active の診断時の動作を変更します。

選択項目▪ Alarm▪ Warning

初期設定

Warning

追加情報

選択項目の詳細については、Diagnostic behavior サブメニューの説明 →を参照してください。

Diagnostic no. 441（Current output 1～n）
ナビゲーション

Expert → System → Diagnostic handling → Diagnostic behavior → Diagnostic no. 441

説明

この機能を使用して、診断メッセージ 441 Current output 1～n の診断時の動作を変更します。

選択項目

▪ Off
▪ Alarm

Endress+Hauser

25

機能説明書

J22 TDLAS ガスアナライザ▪ Warning▪ Logbook entry only
初期設定

Warning

追加情報

選択項目の詳細については、Diagnostic behavior サブメニューの説明 →を参照してください。

Diagnostic no. 444（Current input 1～n）
ナビゲーション

Expert → System → Diagnostic handling → Diagnostic behavior → Diagnostic no. 444

必須条件

機器に 1 つの電流入力があること。

説明

この機能を使用して、診断メッセージ 444 Current input 1～n の診断時の動作を変更します。

選択項目

▪ Off
▪ Alarm▪ Warning▪ Logbook entry only

初期設定

Warning

追加情報

選択項目の詳細な説明：→

31

Diagnostic no. 905（Validation failed）
ナビゲーション

Expert → System → Diagnostic handling → Diagnostic behavior → Diagnostic no. 905

説明

この機能を使用して、診断メッセージ 905 Validation failed の診断時の動作を変更します。

選択項目

▪ Off
▪ Alarm▪ Warning▪ Logbook entry only
▪ Reset

初期設定

Warning

追加情報

選択項目の詳細については、Diagnostic behavior サブメニューの説明 →を参照してくだ

さい。

3.1.4

Administration

ナビゲーション

Expert → System → Administration‣ Administration

26

→

26

Device reset→

27

Transmitter identifier→

27

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Activate SW option→

28

Software option overview→

28‣ Define access code→29

‣ Reset access code→30

Device reset
Expert → System → Administration → Device reset

ナビゲーション
説明

すべてまたは一部の機器設定を所定の状態にリセットします。

選択項目▪ Cancel▪ Restart device▪ To delivery settings▪ Restore S-DAT backup

初期設定

Cancel

追加情報

選択項目

1
3F

▪ Cancel：何も実行せずにこのパラメータを終了します。
▪ Restart device：再起動により、揮発性メモリ（RAM）に保存されているすべてのパラメー
タを初期設定にリセットします（例：測定値データ）。機器設定に変更はありません。
▪ To delivery settings：ユーザー固有の初期設定で注文されたすべてのパラメータをユーザー固有の値にリセットします。その他のパラメータはすべて初期設定にリセットされます。
▪ Restore S-DAT backup：S-DAT に保存されているデータを復元します。追加情報：この機能は、メモリの問題「083 Memory content inconsistent」の解決や、新しい S-DAT を取り付けたときに S-DAT データを復元するために使用できます。
この項目はアラーム状態でのみ表示されます。

Transmitter identifier
Expert → System → Administration → Transmitter identifier

ナビゲーション
説明

変換器の識別名を選択します。

ユーザーインター▪ Unknown

フェース

▪ 500
▪ 300

初期設定

1

300

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser27

機能説明書

J22 TDLAS ガスアナライザ

Activate SW option
Expert → System → Administration → Activate SW option

ナビゲーション
説明

この機能を使用して、注文した追加のソフトウェアオプションを有効にするためのアクティ
ベーションコードを入力します。

ユーザー入力最大 10 桁の数字

初期設定

注文したソフトウェアオプションに応じて異なります。

追加情報

説明
追加ソフトウェアオプション付きで機器を注文した場合、アクティベーションコードは工場
出荷時に機器にプログラムされています。

ユーザー入力後からソフトウェアオプションを有効にする場合は、弊社営業所もしくは販売代理店にお問い合わせください。
不正確なコードまたは無効なコードを入力すると、すでに有効になっているソフトウェアオ
プションが失われます。
▪ 新しいアクティベーションコードを入力する前に、現在のアクティベーションコードを書
き留めておいてください。
▪ 新しいソフトウェアオプションをご注文いただいた場合は、Endress+Hauser が提供する
新しいアクティベーションコードを入力してください。
▪ アクティベーションコードの入力後、新しいソフトウェアオプションが Software option
overview パラメータ →に表示されることを確認してください。

新しいソフトウェアオプションが表示された場合、そのオプションは有効です。
新しいソフトウェアオプションが表示されないか、すべてのソフトウェアオプションが削除された場合、入力されたコードが不正確または無効です。
▪ 入力されたコードが不正確または無効な場合は、古いアクティベーションコードを入力してください。
▪ 弊社営業所もしくは販売代理店に、必ずシリアル番号を提示して新しいアクティベーショ
ンコードの確認を依頼するか、コードを再度要請します。

ソフトウェアオプションの例「Extended HistoROM」
現在有効なソフトウェアオプションが Software option overview パラメータ →

に表示され

ます。

ウェブブラウザソフトウェアオプションを有効にした場合、ウェブブラウザで再度ページの読み込みを行う必要があります。
 アクティベーションコードは、機器のシリアル番号にリンクされており、機器とソフト
ウェアオプションに応じて異なります。

Software option overview
ナビゲーション
説明28

Expert → System → Administration → SW option overview
有効な機器のソフトウェアオプションがすべて表示されます。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書
1

ユーザーインター
フェース▪ Extended HistoROM
1▪ Heartbeat Monitoring
1▪ Heartbeat Verification

追加情報

説明

4F

ユーザーの注文により使用可能なすべてのオプションを表示します。
Define access code ウィザード
Define access code ウィザードは、現場表示器またはウェブブラウザによる操作でのみ使用できます。
操作ツールを使用する場合は、Define access code パラメータが Administration サブメニューに直接表示されます。操作ツールから機器を操作する場合、Confirm access code パラメータはありません。

ナビゲーション

Expert → System → Administration → Define access code
‣ Define access code
Define access code→30

Confirm access code→30

Define access code
Expert → System → Administration → Define access code → Define access code

ナビゲーション
説明

この機能を使用して、パラメータ書き込みアクセスを制限するためのユーザー固有のリリー
スコードを入力します。これにより、現場表示器、ウェブブラウザ、FieldCare、または
DeviceCare（CDI-RJ45 サービスインターフェース経由）を介して機器設定が不用意に変更されることを防止できます。

ユーザー入力数字、英字、特殊文字から成る最大 16 桁の文字列

追加情報

説明
書き込み保護は、本書のシンボルが付いたすべてのパラメータに適用されます。

現場表示器でパラメータの前にシンボルが表示される場合、それが書き込み保護パラ

メータであることを示します。
書き込みアクセスできないパラメータは、ウェブブラウザで灰色表示されます。
 アクセスコードを設定すると、書き込み保護パラメータは Enter access code パラメータ→ でアクセスコードを入力しない限り変更できません。
 アクセスコードを紛失した場合は、弊社営業所もしくは販売代理店にお問い合わせください。

ユーザー入力
アクセスコードが入力レンジを超えた場合はメッセージが表示されます。

初期設定

1

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser

29

機能説明書

J22 TDLAS ガスアナライザ

初期設定を変更していない場合、またはアクセスコードとして 0 を設定している場合、パラメータは書き込み保護されず、機器設定データは変更可能な状態となります。ユーザーはメンテナンスの役割でログオンします。

Confirm access code
Expert → System → Administration → Define access code → Confirm code

ナビゲーション
説明

設定したリリースコードを再度入力して、リリースコードを確定します。

ユーザー入力数字、英字、特殊文字から成る最大 16 桁の文字列

Reset access code サブメニュー

ナビゲーション

Expert → System → Administration → Reset access code
‣ Reset access code
Operating time→

30

Reset access code→30

Operating time
Expert → System → Administration → Reset access code → Operating time

ナビゲーション
説明

この機能を使用して、機器の稼働時間を表示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
追加情報

ユーザーインターフェース
最大日数は 9999 です。これは、27 年に相当します。

Reset access code
Expert → System → Administration → Reset access code → Reset access code

ナビゲーション
説明

この機能を使用して、リセットコードを入力してユーザー固有のアクセスコードを初期設定にリセットします 。

ユーザー入力数字、英字、特殊文字から成る文字列

初期設定

0x00

追加情報

説明
リセットコードについては、Endress+Hauser サービス部門にお問い合わせください。

ユーザー入力リセットコードは、以下を介してのみ入力できます。
▪ ウェブブラウザ▪ フィールドバス
30

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

3.2 Sensor
ナビゲーション

Expert → Sensor‣ Sensor

3.2.1‣ Measured values→

31‣ System units→

38‣ Stream→42‣ Dew point→

43‣ Peak tracking→

44‣ Sensor adjustment→

47‣ Stream change compensation (SCC)→49‣ Calibration→

51‣ Measured variables→

31‣ Input values→

35‣ Output values→

36

Concentration→

32

Dew point 1→

32

Dew point 2→

32

Cell gas pressure→

32

Cell gas temperature→

32

Measured values

ナビゲーション

Expert → Sensor → Measured values‣ Measured values

Measured variables サブメニュー

ナビゲーション

Expert → Sensor → Measured values → Measured variables‣ Measured variables

Endress+Hauser

31

機能説明書

J22 TDLAS ガスアナライザ

Detect. ref. level→

34

Detect. zero level→

34

Peak 1 index→

34

Peak 1 index delta→

34

Peak 2 index→

34

Peak 2 index delta→

34

Peak track index→

34

Peak track index delta→

34

Midpoint delta→

34

Concentration
ナビゲーション
説明

Expert → Sensor → Measured values → Measured variables → Concentration
サンプルセルで現在測定されている被分析物濃度を表示します。

ユーザーインターフェース 0～1000000 ppmv
追加情報

単位は、concentration unit パラメータ →の設定が使用されます。

濃度は、測定するガスサンプル内の気相に含まれる水蒸気量を示します。

Dew point 1
ナビゲーション
必須条件

Expert → Sensor → Measured values → Measured variables → Dew point 1
被分析物タイプが水分「H2O」であること。
Dew point method 1 パラメータで、Off が選択されていないこと。

説明

現在計算されている露点温度を表示します。

ユーザーインターフェース 符号付き浮動小数点数
追加情報

単位は、temperature unit パラメータ →の設定が使用されます。

露点とは所定の濃度と圧力において水分が凝結して液化を開始する温度です。水分の露点計算には各産業分野で認められた複数の方法があります。詳細については、取扱説明書 BA02152C →を参照してください。

Dew point 2
ナビゲーション
必須条件
32

Expert → Sensor → Measured values → Measured variables → Dew point 2
被分析物タイプが水分「H2O」であること。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Dew point method 2 パラメータで、Off が選択されていないこと。
説明

現在計算されている露点温度を表示します。

ユーザーインターフェース 符号付き浮動小数点数
追加情報

単位は、temperature unit パラメータ →の設定が使用されます。

露点とは所定の濃度と圧力において水分が凝結して液化を開始する温度です。水分の露点計算には各産業分野で認められた複数の方法があります。詳細については、取扱説明書 BA02152C →を参照してください。

Cell gas pressure
Expert → Sensor → Measured values → Measured variables → Cell gas pressure

ナビゲーション
説明

サンプルセルで現在測定されているガス圧力を表示します。

ユーザーインターフェース 0～1000000 ppmv
追加情報

単位は、pressure unit パラメータ →の設定が使用されます。

測定中のサンプルセルの現在の圧力

Cell gas temperature
Expert → Sensor → Measured values → Measured variables → Cell gas

ナビゲーション

temperature
説明

サンプルセルで現在測定されているガス温度を表示します。

ユーザーインターフェース 符号付き浮動小数点数
追加情報

単位は、temperature unit パラメータ →の設定が使用されます。

測定中のサンプルセルの現在の温度

Detector reference level
Expert → Sensor → Measured values → Measured variables → Detector reference

ナビゲーション
level
説明

現在測定されているレーザー検出器の基準レベルを表示します。

ユーザーインターフェース 0～5 mA
追加情報

DC レーザー出力の大きさ。値が範囲外である場合、オプティックの洗浄が必要であ
ること、または配置に問題があることを示している可能性があります。

Detector zero level
ナビゲーション
説明

Expert → Sensor → Measured values → Measured variables → Detector zero level
現在測定されているレーザー検出器のゼロレベルを表示します。

ユーザーインターフェース 0～5 mA
追加情報

Endress+Hauser

レーザーがオフのときの DC レーザー出力（例：暗電流）33機能説明書

J22 TDLAS ガスアナライザ

Peak 1 index
Expert → Sensor → Measured values → Measured variables → Peak 1 index

ナビゲーション
説明

現在測定されている 2f スペクトルの吸収ピーク 1 指標位置を表示します。

ユーザーインターフェース 0.0～511.0
追加情報

スキャンに沿った吸収ピークの位置

Peak 1 index delta
Expert → Sensor → Measured values → Measured variables → Peak 1 index delta

ナビゲーション
説明

現在測定されている 2f スペクトルのピーク 1 指標位置とターゲット指標の差を表示します。

ユーザーインターフェース -511.0～+511.0

Peak 2 index
Expert → Sensor → Measured values → Measured variables → Peak 2 index

ナビゲーション
必須条件

2 つのピークに対してアナライザが校正されていること。

説明

現在測定されている 2f スペクトルの吸収ピーク 2 指標位置を表示します。

ユーザーインターフェース 0.0～511.0
追加情報

スキャンに沿った第 2 ピークの位置。ピーク追跡のために使用されます。

Peak 2 index delta
Expert → Sensor → Measured values → Measured variables → Peak 2 index delta

ナビゲーション
必須条件

2 つのピークに対してアナライザが校正されていること。

説明

現在測定されている 2f スペクトルのピーク 2 指標位置とターゲット指標の差を表示します。

ユーザーインターフェース -511.0～+511.0

Peak track index
Expert → Sensor → Measured values → Measured variables → Peak track index

ナビゲーション
説明

現在測定されている 2f スペクトルのピーク追跡に使用されるピークのピーク追跡指標を表示します。

ユーザーインターフェース 0.0～511.0
追加情報

説明
Peak track analyzer control パラメータで Off を選択した場合、この値はゼロになります。それ以外の場合は、この値はピーク追跡に使用されるピークに応じて、Peak
1～n index パラメータと同じになります。

34

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Peak track index delta
Expert → Sensor → Measured values → Measured variables → Peak track index

ナビゲーション
delta

現在測定されている 2f スペクトルのピーク追跡指標とターゲット指標の差を表示し説明ます。
ユーザーインターフェース -511.0～+511.0

説明

追加情報

Peak track analyzer control パラメータで Off を選択した場合、この値はゼロになります。それ以外の場合は、この値はピーク追跡に使用されるピークに応じて、Peak
1～n index delta パラメータと同じになります。

Midpoint delta
Expert → Sensor → Measured values → Measured variables → Midpoint delta

ナビゲーション

校正された中心点値と現在使用されている中心点値の差を表示します。

説明

ユーザーインターフェース 0.0～120.0 mA

説明

追加情報

Peak track analyzer control パラメータで Off を選択した場合、この値はゼロになります。それ以外の場合は、この値はピーク追跡アルゴリズムによって校正された中心点値に適用される変化量になります。
Input values サブメニュー

ナビゲーション

Expert → Sensor → Measured val. → Input values‣ Input values‣ Current input 1～n→35‣ Value status input 1～n→36‣ Measured values 1～n→35‣ Measured current 1～n→36

Current input 1～n サブメニュー

ナビゲーション

Expert → Sensor → Measured val. → Input values → Current input 1～n‣ Current input 1～n

Measured values 1～n
ナビゲーション

Expert → Sensor → Measured values → Input values → Current input 1～n →Measured values 1～nEndress+Hauser

35

機能説明書

J22 TDLAS ガスアナライザ

現在の電流入力値を表示します。

説明

ユーザーインターフェース 符号付き浮動小数点数

Measured current 1～nExpert → Sensor → Measured values → Input values → Current input 1～n →ナビゲーション

Measured current 1～n
電流入力の現在値を表示します。

説明

ユーザーインターフェース 0～22.5 mA

Value status input 1～n サブメニュー

ナビゲーション

Expert → Sensor → Measured values → Input values → Value status input 1～n‣ Value status current input 1～n‣ Value status input→

36

Value status input
Expert → Sensor → Measured values → Input values → Value status input 1～n →ナビゲーション

Value status input
現在の入力信号レベルを表示

説明

ユーザーインターフェース ▪ High▪ Low
Output values サブメニュー

ナビゲーション

Expert → Sensor → Measured val. → Output values‣ Output values‣ Value current output 1～n→36‣ Switch output 1～n→37‣ Relay output 1～n

→

37

→

37

Value current output 1～n サブメニュー

ナビゲーション

Expert → Sensor → Measured val. → Output values → Val. curr.outp 1～n‣ Value current output 1～n
Output current 1～n36Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Measured current 1～n→37

Output current 1～n
ナビゲーション

Expert → Sensor → Measured values → Output values → Value current output 1～n →Output current 1～n

説明

現在計算されている電流出力の電流値を表示します。

ユーザーインターフェース 0～22.5 mA
Measured current 1～n
ナビゲーション

Expert → Sensor → Measured val. → Output values → Val. Current output 1～n →Measured current 1～n

説明

電流出力の実際の測定値を表示します。

ユーザーインターフェース 0～30 mA
Switch output 1～n サブメニュー

ナビゲーション

Expert → Sensor → Measured val. → Output values → Switch output 1～n‣ Switch output 1～n
Switch state 1～n→37

Switch state 1～n
ナビゲーション

Expert → Sensor → Measured val. → Output values → Switch output 1～n →Switch state 1～n

必須条件

Operating mode パラメータ →説明ステータス出力の現在のステータス切り替えを表示します。

で、Switch が選択されていること。

ユーザーインターフェース ▪ Open▪ Closed
追加情報

ユーザーインターフェース▪ Open：スイッチ出力は非導通です。
▪ Closed：スイッチ出力は導通です。

Relay output 1～n サブメニュー

ナビゲーション

Expert → Sensor → Measured val. → Output values → Relay output 1～n‣ Relay output 1～n
Switch state

Endress+Hauser

→

38
37

機能説明書

J22 TDLAS ガスアナライザ

Switch cycles→

38

Max. switch cycles number→

38

Switch state
ナビゲーション

Expert → Sensor → Measured val. → Output values → Relay output 1～n → Switch state説明

リレー出力の現在のステータスを表示します。

ユーザーインターフェース ▪ Open▪ Closed

ユーザーインターフェース

追加情報▪ Open：リレー出力は非導通です。
▪ Closed：リレー出力は導通です。
Switch cycles
ナビゲーション

Expert → Sensor → Measured val. → Output values → Relay output 1～n → Switch
cycles

説明

実行されたすべてのスイッチサイクルを表示します。

ユーザーインターフェース 正の整数
Max. switch cycles number
ナビゲーション

Expert → Sensor → Measured val. → Output values → Relay output 1～n → Max. cycles no.

説明

保証されるスイッチサイクルの最大数を表示します。

ユーザーインターフェース 正の整数

3.2.2

System units

ナビゲーション

Expert → Sensor → System units‣ System units

38

Concentration unit (ppmv)→

39

Temperature unit (°C)→

39

Pressure unit (bar)→40

Length unit (m)→40Date/time format→

40

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書‣ User-specific units→

41

Concentration unit
ナビゲーション

Expert → Sensor → System units → Concentration unit

説明

この機能を使用して、濃度の単位を選択します。

選択項目▪ ppmv▪ ppbv▪ %vol▪ lb/MMscf▪ mg/sm3▪ mg/Nm3▪ User conc.

初期設定

ppmv

追加情報

影響
選択した単位は以下に適用：▪ Concentration パラメータ →▪ Concentration offset →▪ Validation concentration →▪ Measured concentration →▪ Concentration average →▪ Concentration standard deviation →▪ Concentration minimum →▪ Concentration maximum →選択項目
単位の短縮表記の説明については、認定固有の初期設定 →を参照してください。

Temperature unit
ナビゲーション

Expert → Sensor → System units → Temperature unit

説明

この機能を使用して、温度の単位を選択します。

選択項目

SI 単位

US 単位

▪ °C

▪ °F

▪ K

▪ °R

初期設定

認定固有の項目：
▪ °C
▪ °F

追加情報

影響
選択した単位は以下に適用：▪ Cell gas temperature →▪ Dew point 1 パラメータ →▪ Dew point 2 パラメータ →Endress+Hauser

39

機能説明書

J22 TDLAS ガスアナライザ

選択項目
単位の短縮表記の説明については、認定固有の初期設定 →を参照してください。

Pressure unit
ナビゲーション

Expert → Sensor → System units → Pressure unit

説明

この機能を使用して、プロセス圧力の単位を選択します。

選択項目

SI 単位

US 単位

▪ MPa a▪ psi a▪ MPa g▪ psi g▪ kPa a▪ kPa g▪ Pa a▪ Pa g▪ bar▪ bar g
初期設定

認定固有の項目：▪ bar a▪ psi a

追加情報

結果
単位は以下の設定が使用されます。
▪ Cell gas pressure value パラメータ →▪ Pipeline pressure fixed →▪ Pipeline pressure →選択項目
単位の短縮表記の説明については、認定固有の初期設定 →を参照してください。

Length unit
ナビゲーション

Expert → Sensor → System units → Length unit

説明

この機能を使用して、呼び口径の長さの単位を選択します。

選択項目

▪ m
▪ ft
▪ in
▪ mm
▪ µm

初期設定

m

追加情報

選択項目
単位の短縮表記の説明については、認定固有の初期設定 →40を参照してください。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Date/time format
ナビゲーション

Expert → Sensor → System units → Date/time format

説明

この機能を使用して、必要な校正履歴の時刻フォーマットを選択します。

選択項目▪ dd.mm.yy hh:mm▪ dd.mm.yy hh:mm am/pm▪ mm/dd/yy hh:mm▪ mm/dd/yy hh:mm am/pm

初期設定

dd.mm.yy hh:mm

追加情報

選択項目
単位の短縮表記の説明については、認定固有の初期設定 →を参照してください。

User-specific units サブメニュー

ナビゲーション

Expert → Sensor → System units → User-specific units‣ User-specific units
User concentration text→

41

User concentration offset→

41

User concentration factor→

41

User concentration text
ナビゲーション
説明

Expert → Sensor → System units → User-specific units → User concentration text
この機能を使用して、ユーザー固有の濃度単位のテキストを入力します。
対応する濃度単位が自動的に生成されます。

ユーザー入力最大 10 文字（英字、数字、特殊文字（@、%、/）など）

初期設定

User conc.

追加情報

結果
設定した単位は、concentration unit パラメータ →れます。

の選択リストに選択項目として表示さ例
質量百万分率の単位としてテキスト「ppmw」を入力します。
User concentration offset
ナビゲーション

Expert → Sensor → System units → User-specific units → User concentration offset

説明

この機能を使用して、ユーザー固有の濃度単位のゼロ点シフトを入力します。

ユーザー入力

符号付き浮動小数点数

Endress+Hauser

41

機能説明書

J22 TDLAS ガスアナライザ

初期設定

0.0

追加情報

ユーザー固有の単位の値 = (係数 × 基本単位の値) + オフセット

User concentration factor
ナビゲーション

Expert → Sensor → System units → User-specific units → User concentration factor

説明

この機能を使用して、ユーザー固有の濃度単位の係数を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

1.0

3.2.3

Streamナビゲーション

Expert → Sensor → Stream‣ Stream
Analyte type→42

Select calibration→

42

Rolling average number→

42

Analyte type
ナビゲーション

Expert → Sensor → Stream → Analyte type

説明

アナライザの校正対象となる被分析物を表示します。

ユーザーインター
フェース

▪ H2O
▪ CO2
▪ H2S
▪ CH4
▪ NH3
▪ HCl
▪ O2
▪ CO
▪ SO2
▪ C2H2

Select calibration
ナビゲーション
説明42

Expert → Sensor → Stream → Select calibration
測定に使用する校正を選択しますアナライザに対して複数の校正を選択できます。
Endress+Hauser

J22 TDLAS ガスアナライザ

選択項目

機能説明書

▪ 1
▪ 2
▪ 3
▪ 4

初期設定

1

追加情報

一部のアナライザでは、検証ガスの校正を含む複数の校正による設定が可能です。ガス流の校正については、付属の校正レポートを参照してください。

Rolling average number
ナビゲーション
説明

Expert → Sensor → Stream → Rolling average number
移動平均に含める濃度測定回数を表示します。

ユーザーインターフェース 1～256

3.2.4

Dew point

ナビゲーション

Expert → Sensor → Dew point‣ Dew point
Dew point method 1→

43

Dew point method 2→

43

Conversion type→43

Pipeline pressure mode→

43

Pipeline pressure fixed→

44

Pipeline pressure→

44‣ Calibration 1～n→45

Dew point method 1
ナビゲーション

Expert → Sensor → Dew point → Dew point method 1

説明

濃度と圧力からの変換に使用する露点温度の計算方法を選択します。

選択項目

▪ Off
▪ ASTM1▪ ASTM2
▪ ISO
▪ AB

初期設定
Endress+Hauser

Off
43機能説明書

J22 TDLAS ガスアナライザ

Dew point method 2
ナビゲーション

Expert → Sensor → Dew point → Dew point method 2

説明

濃度と圧力からの変換に使用する露点温度の計算方法を選択します。

選択項目

▪ Off
▪ ASTM1▪ ASTM2
▪ ISO
▪ AB

初期設定

Off

Conversion type
ナビゲーション

Expert → Sensor → Dew point → Conversion type

説明

露点計算に理想気体と実在気体のいずれの原理を使用するかを選択します。

選択項目▪ Ideal▪ Real

初期設定

Ideal

Pipeline pressure mode
ナビゲーション

Expert → Sensor → Dew point → Pipeline pressure mode

説明

サンプル配管内圧力の入力方法を選択します。

選択項目▪ Fixed value▪ External value

初期設定

Fixed

Pipeline pressure fixed
ナビゲーション

Expert → Sensor → Dew point → Pipeline pressure fixed

説明

サンプル配管内圧力の固定値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

0.0000 bar

Pipeline pressure external
ナビゲーション

Expert → Sensor → Dew point → Pipeline pressure external

説明

サンプル配管内圧力の外部値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

0.0000 bar

44

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Calibration 1～n サブメニュー

ナビゲーション

Expert → Sensor → Dew point → Calibration 1～n‣ Calibration 1～nMethane CH4→

45

Ethane C2H6→45

Propane C3H8→45

IButane C4H10→45

N-Butane C4H10→45

Isopentane C5H12→45

N-Pentane C5H12→45

Neopentane C5H12→45

Hexane+ C6H14+→

45

Nitrogen N2→

45

Carbon diox. CO2→

45

Hydrog.sulf. H2S→

45

Hydrogen H2→

45

Component (n)ナビゲーション
説明

Expert → Sensor → Dew point → Calibration 1～n → Component (n)ガス流内の各バックグラウンド成分のモル分率を表します。
下表内の「mol」は「モル分率」の略語です。

Endress+Hauser

パラメータ

説明

ユーザー入力

初期設定

Stream change
compensation

ガス流変化の補正機能を有効化/無効化しま
す。

• On

Off

Methane CH4

ドライガス混合物のメタンのモル分率を設定します。

0.4～1.0 mol

0.75 molEthane C2H6

ドライガス混合物のエタンのモル分率を設定します。

0.0～0.2 mol

0.1 mol

Propane C3H8

ドライガス混合物のプロパンのモル分率を設定します。

0.0～0.15
mol

0.05 molIButane
C4H10

ドライガス混合物の i-ブタンのモル分率を設定します。

0.0～0.1 mol

0 mol

• Off

45

機能説明書

J22 TDLAS ガスアナライザ

パラメータ

説明

ユーザー入力

初期設定

N-Butane
C4H10

ドライガス混合物の n-ブタンのモル分率を設定します。

0.0～0.1 mol

0 mol

Isopentane
C5H12

ドライガス混合物のイソペンタンのモル分率を設定します。

0.0～0.1 mol

0 mol

N-Pentane
C5H12

ドライガス混合物の n-ペンタンのモル分率を設定します。

0.0～0.1 mol

0 mol

Neopentane
C5H12

ドライガス混合物のネオペンタンのモル分率を設定します。

0.0～0.1 mol

0 mol

Hexane+C6H14+

ドライガス混合物のヘキサン+ のモル分率を設定します。

0.0～0.1 mol

0 mol

Nitrogen N2

ドライガス混合物の窒素のモル分率を設定します。

0.0～0.55
mol

0 mol

Carbon dioxide
CO2

ドライガス混合物の二酸化炭素のモル分率を設定します。

0.0～0.3 mol

0.1 molHydrogensulfide H2S

ドライガス混合物の硫化水素のモル分率を設定します。

0.0～0.05
mol

0 mol

Hydrogen H2

ドライガス混合物の水素のモル分率を設定します。

0.0～0.2 mol

0 mol

ユーザー入力正の浮動小数点値（上記の表の各成分を参照）

初期設定

表を参照

追加情報

各バックグラウンド成分のモル分率の合計が 1 になる必要があります。

3.2.5

Peak tracking

ナビゲーション

Expert → Sensor → Peak tracking‣ Peak tracking
Peak track analyzer control→

46

Peak track reset→

46

Peak track average number→

46

Peak track analyzer control
ナビゲーション
説明

Expert → Sensor → Peak tracking → Peak track analyzer control
アナライザのピーク追跡のオン/オフを切り替えます。校正ごとに個別のピーク追跡設定があります。通常動作では、ピーク追跡をオンにする必要があります。

46

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

選択項目

▪ Off
▪ On

初期設定

Off

Peak track reset
ナビゲーション

Expert → Sensor → Peak tracking → Peak track reset

説明

アナライザのピーク中心点の現在値を最初の校正後のピーク位置にリセットします。

選択項目

▪ Off
▪ Reset

初期設定

Off

Peak track average number
ナビゲーション

Expert → Sensor → Peak tracking → Peak track average number

説明

ピーク追跡に使用するピーク指標の平均測定回数

ユーザー入力1～3600

初期設定

60

3.2.6

Sensor adjustment

ナビゲーション

Expert → Sensor → Sensor adjustment‣ Sensor adjustment
Concentration adjust→

47

Conc. multiplier→

47

Concentration offset(RATA)→

47

2f base crv source→

47

2f base RT update→

49

Calibration 1～n→48

Concentration adjust
ナビゲーション

Expert → Sensor → Sensor adjustment → Concentration adjust

説明

濃度調整機能（例：濃度乗数、濃度オフセット）のオン/オフを切り替えます。

選択項目

▪ On
▪ Off

Endress+Hauser

47

機能説明書

J22 TDLAS ガスアナライザ

初期設定

Off

追加情報

工場校正に影響を与えずに、ユーザーがアナライザの測定値調整を定義できます。

Conc. multiplier
ナビゲーション

Expert → Sensor → Sensor adjustment → Conc. multiplier

説明

濃度調整機能がオンのときに濃度に乗じる値を設定します。

ユーザーインター

符号付き浮動小数点数

フェース
初期設定

1.0000Concentration offset (RATA)
ナビゲーション

Expert → Sensor → Sensor adjustment → Concentration offset (RATA)

説明

濃度調整機能がオンのときに濃度に加算する値（オフセット）を設定します。

ユーザーインター

符号付き浮動小数点数

フェース
初期設定

0.0000 ppmv

2f base curve source
ナビゲーション
説明

Expert → Sensor → Sensor adjustment → 2f base curve source
測定の校正に使用するベース曲線のソース（工場校正の Ref0 または最後の RT 更新のRef0）を選択します。

選択項目▪ Ref0 curve▪ Ref0 RT curve

初期設定

Ref0 curve

2f base RT update
ナビゲーション
説明

Expert → Sensor → Sensor adjustment → 2f base curve source
Ref0 RT curve を選択した場合に Start を選択すると、測定値計算用の RT（リアルタイム）
ベース曲線データの保存が開始されます。

選択項目▪ Cancel▪ Start

初期設定

48

Ref0 curve

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Calibration 1～n サブメニュー

ナビゲーション

Expert → Sensor → Sensor adjustment → Calibration 1～n‣ Calibration 1
Laser midpoint default→

48

Laser ramp default→

48

Mod amp default→

48

Laser midpoint default
ナビゲーション
説明

Expert → Sensor → Sensor adjustment → Calibration 1～n → Laser midpoint default
各校正ガスの工場校正された中心点を表示します。

ユーザーインターフェース 0～120 mA
追加情報

この値は、ピーク位置を最適化するための中心点デルタの始点として利用できます。

Laser ramp default
ナビゲーション
説明

Expert → Sensor → Sensor adjustment → Calibration 1～n → Laser ramp default
各校正ガスの工場校正されたランプを表示します。

ユーザーインターフェース 0～120 mA
追加情報

レーザーランプはスペクトルのスキャン幅を示します。

Laser modulation amplitude default
ナビゲーション
説明

Expert → Sensor → Sensor adjustment → Calibration 1～n → Mod amp default
ピーク性能を最適化するための振幅変調の設定

ユーザーインターフェース 0～100 mA

3.2.7

Stream change compensation

ナビゲーション

Expert → Sensor → Stream change compensation‣ Stream change compensation (SCC)‣ Calibration 1～nStream change
compensation

Endress+Hauser

→

49

→

51

49

機能説明書

J22 TDLAS ガスアナライザ

Methane CH4→51

Ethane C2H6→51

Propane C3H8→

51

IButane C4H10→

51

N-Butane C4H10→51

Isopentane C5H12→

51

N-Pentane C5H12→51

Neopentane C5H12→

51

Hexane+ C6H14+→

51

Nitrogen N2→

51

Carbon diox. CO2→

51

Hydrog.sulf. H2S→

51

Hydrogen H2→

51

Calibration 1～n → Stream change compensation
Expert → Sensor → Stream change compensation → Calibration 1～n → Stream change

ナビゲーション

compensation
説明

ガスのバックグラウンド組成値に基づく濃度測定補正機能をオンにします。値は固定値または実際の値を使用できます。

選択項目

▪ Off
▪ On

初期設定

Off

Component (n)ナビゲーション

Expert → Sensor → Stream change compensation → Calibration 1～n → Component (n)説明

これらの値により、ガスのバックグラウンド値を設定します。値は露点と共有されます。

ユーザー入力

符号付き浮動小数点数、モル分率

初期設定

ガスのバックグラウンドに依存します。露点校正成分 →50を参照してください。

Endress+Hauser

J22 TDLAS ガスアナライザ

3.2.8

機能説明書

Calibration

ナビゲーション

Expert → Sensor → Calibration‣ Calibration
Det. 1 TIA gain→51

Detector bias→

51

Flow switch input→

51

Flow switch state→

51

Det. 1 TIA gain
ナビゲーション

Expert → Sensor → Calibration → Det. 1 TIA gain

説明

トランスインピーダンスアンプ（TIA）のゲイン設定

選択項目

0～15

Detector bias
ナビゲーション

Expert → Sensor → Calibration → Detector bias

説明

光学式検出器の運転に使用するバイアス電圧

選択項目

符号付き浮動小数点数

Flow switch input
ナビゲーション

Expert → Sensor → Calibration → Flow switch input

説明

サンプルガスの流量あり/流量なしを通知するフロースイッチのディスクリート入力

選択項目▪ Normally Open▪ Normally Closed▪ Off

Flow switch state
ナビゲーション

Expert → Sensor → Calibration → Flow switch state説明

フロースイッチの現在のステータスを表示します。

選択項目▪ No Flow▪ Flow

Endress+Hauser

51

機能説明書

J22 TDLAS ガスアナライザ

3.3 I/O configuration
ナビゲーション

Expert → I/O configuration‣ I/O configuration
I/O module 1～n terminal numbers→

52

I/O module 1～n information→

52

I/O module 1～n type→52

Apply I/O configuration→

53

I/O alteration code→53

I/O module 1～n terminal numbers
ナビゲーション

Expert → I/O configuration → I/O module 1～n terminals

説明I/O モジュールが使用している端子番号を表示します。

ユーザーインター▪ Not usedフェース

▪ 26-27（I/O 1）▪ 24-25（I/O 2）▪ 22-23（I/O 3）

I/O module 1～n information
ナビゲーション

Expert → I/O configuration → I/O module 1～n information

説明

接続された I/O モジュールに関する情報を表示します。

ユーザーインター▪ Not plugged

フェース▪ Invalid▪ Not configurable▪ Configurable▪ MODBUS

追加情報▪ Not plugged：I/O モジュールが接続されていません。
▪ Invalid：I/O モジュールが正しく接続されていません。
▪ Not configurable：I/O モジュールは設定できません。
▪ Configurable：I/O モジュールは設定可能です。
▪ MODBUS：I/O モジュールは Modbus 用に設定されています。

52

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

I/O module 1～n type
Expert → I/O configuration → I/O module 1～n type

ナビゲーション
必須条件

I/O モジュールが設置されていること。次のオーダーコードの場合：▪ 「出力；入力 2」、「設定可能 I/O の初期設定オフ」▪ 「出力；入力 3」、「設定可能 I/O の初期設定オフ」

説明

この機能を使用して、I/O モジュールの設定のための I/O モジュールタイプを選択します。

選択項目

▪ Off
▪ Current output▪ Current input

1▪ Switch output

1▪ Relay output
初期設定

1
5F

1

Off

Apply I/O configuration
Expert → I/O configuration → Apply I/O configuration

ナビゲーション
説明

この機能を使用して、新たに設定した I/O モジュールタイプを有効にします。

選択項目

▪ No
▪ Yes

初期設定

No

I/O alteration code
Expert → I/O configuration → I/O alteration code

ナビゲーション
説明

各 I/O の設定を有効化します。

ユーザー入力

正の整数

初期設定

機器固有

追加情報

説明
I/O 設定は、I/O module type パラメータ →1で変更します。

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser

53

機能説明書

J22 TDLAS ガスアナライザ

3.4 Input
ナビゲーション

Expert → Input‣ Input‣ Current input 1～n

3.4.1

→

54

Terminal number→

54

Signal mode→

54

Current span→55

0/4 mA value→

55

20 mA value→

55

Failure mode→

56

Failure value→

56

Current input 1～n

ナビゲーション

Expert → Input → Current input 1～n‣ Current input 1～n

Terminal number
ナビゲーション

Expert → Input → Current input 1～n → Terminal number

説明

電流入力モジュールが使用している端子番号を表示します。

ユーザーインター
フェース▪ Not used
▪ 24-25（I/O 2）▪ 22-23（I/O 3）

追加情報「Not used」オプション
電流入力モジュールは端子番号を使用していません。

Signal mode
ナビゲーション

Expert → Input → Current input 1～n → Signal mode

説明

この機能を使用して、電流入力の信号モードを選択します。

ユーザーインター
フェース▪ Passive▪ Active

追加情報

Active54

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Current span
Expert → Input → Current input 1～n → Current span

ナビゲーション
説明

この機能を使用して、プロセス値出力の電流範囲とアラーム時の信号の上限/下限レベルを選択します。

選択項目▪ 0-20 mA▪ 4-20 mA NAMUR▪ 4-20 mA US▪ FIXED CURRENT

初期設定

認定固有の項目：▪ 4 to 20 mA NAMUR (3.8 to 20.5 mA)▪ 4 to 20 mA US (3.9 to 20.8 mA)

追加情報

電流範囲のサンプル値：Current range output →.

0/4 mA value
Expert → Input → Current input 1～n → 0/4 mA value

ナビゲーション
説明

この機能を使用して、4 mA の値を入力します。

選択項目

符号付き浮動小数点数

初期設定

0

追加情報

電流入力の挙動
電流入力の挙動は、以下のパラメータの設定に応じて異なります。
▪ Current span →▪ Failure mode →設定例
4 mA value パラメータ →の設定例に注意してください。

20 mA value
ナビゲーション

Expert → Input → Current input 1～n → 20 mA value

説明

この機能を使用して、20 mA の値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

国および工場校正に応じて異なります。

追加情報

設定例
4 mA value パラメータ →Endress+Hauserの設定例に注意してください。

55

機能説明書

J22 TDLAS ガスアナライザ

Failure mode
Expert → Input → Current input 1～n → Failure mode

ナビゲーション
説明

この機能を使用して、設定された Current span パラメータ →の範囲外の電流を測定した

ときの入力動作を選択します。
選択項目▪ Alarm▪ Last valid value▪ Defined value

初期設定

Alarm

追加情報

選択項目
▪ Alarm：エラーメッセージが設定されます。
▪ Last valid value：最後の有効な測定値が使用されます。
▪ Defined value：Failure value パラメータ →Failure value
Expert → Input → Current input 1～n → Failure value

ナビゲーション
必須条件

Failure mode パラメータ →で、Defined value が選択されていること。

説明

この機能を使用して、機器が外部機器から入力信号を受信しない場合、または入力信号が無効な場合に機器が使用する値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

0

56

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

3.5 Output
ナビゲーション

Expert → Output‣ Output

3.5.1‣ Current output 1～n→

57‣ Switch output 1～n→62‣ Relay output 1～n→67

Terminal number→

57

Signal mode→

63

Process variable current output→

63

Current range out→

63

Fixed current→

59

Lower range value output→

59

Upper range value output→

60

Damping current output→

60

Failure behavior current output→

61

Failure current→

61

Output current 1～n→

61

Measured current 1～n→61

Current output 1～n

ナビゲーション

Expert → Output → Current output 1～n‣ Current output 1～n

Terminal number
ナビゲーション
説明

Endress+Hauser

Expert → Output → Current output 1～n → Terminal number
電流出力モジュールが使用している端子番号を表示します。

57

機能説明書

J22 TDLAS ガスアナライザ

ユーザーインター
フェース▪ Not used
▪ 24-25（I/O 2）▪ 22-23（I/O 3）

追加情報「Not used」オプション
電流出力モジュールは端子番号を使用していません。

Signal mode
ナビゲーション

Expert → Output → Current output 1～n → Signal mode

説明

この機能を使用して、電流出力の信号モードを選択します。

選択項目▪ Active▪ Passive

初期設定

ActiveProcess variable current output
ナビゲーション

Expert → Output → Current output 1～n → Process variable current output

説明

この機能を使用して、電流出力に割り当てるプロセス変数を選択します。

選択項目

▪ Off
▪ Concentration▪ Dew Point 1▪ Dew Point 2▪ Cell Gas Temperature

Current range output
ナビゲーション

Expert → Output → Current output 1～n → Current range output

説明

プロセス値出力の電流範囲とアラーム信号の上限/下限レベルを選択します。

選択項目▪ 0-20 mA▪ 4-20 mA NAMUR▪ 4-20 mA US▪ FIXED CURRENT

初期設定

認定固有の項目：▪ 4...20 mA NAMUR (3.8. 20.5 mA)▪ 4...20 mA US (3.9. 20.8 mA)

追加情報

説明
▪ 機器アラームが発生した場合、電流出力は Failure mode パラメータ → で指定された値を取ります。
▪ 測定値が測定範囲外である場合、診断メッセージ S441 Current output 1～n が表示されます。
▪ 測定範囲は、Lower range value output パラメータ → と Upper range value output パラ
メータ →で指定します。

「Fixed current」オプション
電流値は、Fixed current パラメータ →58

で設定します。
Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

例
プロセス値出力用の電流範囲とアラーム時の信号の 2 つのレベルの関係を以下に示します。

I [mA]

A0034351

1.
2.
3.

プロセス値の電流範囲
アラーム時の信号の下限レベル
アラーム時の信号の上限レベル

選択項目

1

2

3

4...20 mA NAMUR (3.8…20.5 mA)

3.8～20.5 mA< 3.6 mA

> 21.95 mA

4...20 mA US (3.9…20.8 mA)

3.9～20.8 mA US< 3.6 mA

> 21.95 mA

4...20 mA (4…20.5 mA)

4～20.5 mA< 3.6 mA

> 21.95 mA

0...20 mA (0…20.5 mA)

0～20.5 mA

0 mA> 21.95 mA

測定値がアラーム時の信号の上限レベルを上回るか、または下限レベルを下回った場合、診断メッセージ

S441 Current output 1～n が表示されます。

Fixed Current
Expert → Output → Current output 1～n → Fixed current

ナビゲーション
必須条件

Current range output パラメータ →で、Fixed current が選択されていること。

説明

この機能を使用して、一定の電流出力値を入力します。

ユーザー入力0～22.5 mA

初期設定

22.5 mA

Lower range value output
ナビゲーション
必須条件

Expert → Output → Current output 1～n → Lower range output
Current range output パラメータ →で、以下の選択項目のいずれかが選択されていること。

▪ 0-20 mA▪ 4-20 mA NAMUR▪ 4-20 mA US▪ FIXED CURRENT
説明

この機能を使用して、測定範囲の開始値を入力します。

ユーザー入力非負の符号付き浮動小数点数

初期設定

0 ppmv

追加情報

依存関係

Endress+Hauser

単位については、Assign current output パラメータ →で選択したプロセス変数に従います。
59

機能説明書

J22 TDLAS ガスアナライザ

電流出力挙動
電流出力の挙動は、以下のパラメータの設定に応じて異なります。
• Current span →• Failure mode →Upper range value output
Expert → Output → Current output 1～n → Upper range output

ナビゲーション
必須条件

Current range output →で、以下の選択項目のいずれかが選択されていること。

▪ 0-20 mA▪ 4-20 mA NAMUR▪ 4-20 mA US▪ FIXED CURRENT
説明

この機能を使用して、測定範囲の終了値を入力します。

ユーザー入力正の符号付き浮動小数点数

初期設定
追加情報

依存関係
単位については、Assign current output パラメータ →で選択したプロセス変数に従います。

Damping current output
Expert → Output → Current output 1～n → Damping current output

ナビゲーション
必須条件

Assign current output パラメータ →output →でプロセス変数が選択されており、Current range

で以下の選択項目のいずれかが選択されていること。

▪ 0-20 mA▪ 4-20 mA NAMUR▪ 4-20 mA US▪ FIXED CURRENT
説明

この機能を使用して、プロセス条件に起因する測定値の変動に対する、電流出力信号の応答
時間の時定数を入力します。

ユーザー入力0.0～999.9 s

初期設定

1.0 s

追加情報

この機能を使用して、電流出力のダンピングの時定数（PT1 エレメント ）を入力します。

1▪ 小さな時定数を入力した場合、電流出力が変動する測定変数に対して非常に素早く反応します。
▪ 一方、大きな時定数を入力した場合は、電流出力の反応が遅くなります。
0 を入力するとダンピングはオフになります（初期設定）。

1

1 次遅延を伴う比例伝送特性

60Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Failure behavior current output
Expert → Output → Current output 1～n → Failure behavior output

ナビゲーション
必須条件

Assign current output パラメータ →output →でプロセス変数が選択されており、Current range

で以下の選択項目のいずれかが選択されていること。

▪ 0-20 mA▪ 4-20 mA NAMUR▪ 4-20 mA US▪ FIXED CURRENT
説明

この機能を使用して、機器アラーム発生時の電流出力値を選択します。

選択項目▪ Min.
▪ Max.
▪ Last valid value▪ Actual value▪ Fixed value

初期設定

Max.

追加情報

説明
この設定は、他の出力のフェールセーフモードには影響しません。これは別のパラメータで設定されます。

「Min.」オプション
電流出力はアラーム時の信号の下限レベルの値を取ります。
アラーム時の信号のレベルは、Current range output →で設定します。

「Max.」オプション
電流出力はアラーム時の信号の上限レベルの値を取ります。
アラーム時の信号のレベルは、Current range output →で設定します。

「Last valid value」オプション
エラー発生時直前に有効だった最後の測定値を出力します。

「Actual value」オプション
電流出力は現在の測定に基づいた測定値を取ります。機器アラームは無視されます。

「Defined value」オプション
設定した測定値を出力します。
この測定値は、Failure current パラメータ →で設定します。

Failure current
ナビゲーション

Expert → Output → Current output 1～n → Failure current

必須条件

Failure mode パラメータ →説明

この機能を使用して、機器アラーム発生時の電流出力の固定値を入力します。

Endress+Hauser

で、Defined value が選択されていること。

61

機能説明書

J22 TDLAS ガスアナライザ

ユーザー入力0～22.5 mA

初期設定

22.5 mA

Output current 1～nExpert → Output → Current output 1～n → Output current 1～n

ナビゲーション

現在計算されている電流出力の電流値を表示します。

説明

ユーザーインターフェース 0～22.5 mA

Measured current 1～n
ナビゲーション

Expert → Output → Current output 1～n → Measured current 1～n電流出力の実際の測定値を表示します。

説明

ユーザーインターフェース 0～30 mA

3.5.2

Switch output 1

ナビゲーション

Expert → Output → Switch output 1～n‣ Switch output 1～n62

Signal mode→

63

Operating mode→

63

Switch out funct→

63

Assign diagnostic behavior→

63

Assign limit→

64

Switch-on value→

64

Switch-off value→

65

Assign status→65

Switch-on delay→

65

Switch-off delay→

65

Switch state→

65

Invert output signal→

65

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Signal mode
ナビゲーション

Expert → Output → Switch output 1～n → Signal mode

説明

この機能を使用して、スイッチ出力の信号モードを選択します。

選択項目▪ Passive▪ パッシブ NAMUR

追加情報▪ Passive▪ Active

Operating mode
ナビゲーション

Expert → Output → Switch output 1～n → Operating mode

説明

出力の動作モードを表示します。

選択項目

Switch

初期設定

SwitchSwitch output function
ナビゲーション

Expert → Output → Switch output 1～n → Switch out function

必須条件

Operating mode パラメータ →で、Switch が選択されていること。

説明

この機能を使用して、スイッチ出力の機能を選択します。

選択項目

▪ Off
▪ On
▪ Diagnostic behavior▪ Limit▪ Status

初期設定

Off

追加情報

選択項目
▪ Off：スイッチ出力が恒久的にオフになります（オープン、非導通）。
▪ On：スイッチ出力が恒久的にオンになります（クローズ、導通）。
▪ Diagnostic behavior：診断イベントの有無を示します。これは診断情報を出力し、システ
ムレベルで適切に反応するために使用されます。
▪ Limit：プロセス変数が規定されたリミット値に達したことを示します。これはプロセスに関する診断情報を出力し、システムレベルで適切に反応するために使用されます。
▪ Status：検証制御が選択された場合に機器ステータスを表示します。

Assign diagnostic behavior
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Assign diagnostic behavior▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →Endress+Hauser

で、Diagnostic behavior が選択されていること。
63機能説明書

J22 TDLAS ガスアナライザ

説明

この機能を使用して、スイッチ出力用に表示される診断イベントカテゴリを選択します。

選択項目▪ Alarm▪ Alarm or warning▪ Warning

初期設定

Alarm

追加情報

説明
診断イベントが未処理の場合、スイッチ出力はクローズ/導通となります。

選択項目▪ Alarm：スイッチ出力はアラームカテゴリの診断イベントのみを通知します。
▪ Alarm or warning：スイッチ出力はアラームおよび警告カテゴリの診断イベントを通知します。
▪ Warning：スイッチ出力は警告カテゴリの診断イベントのみを通知します。
Assign limit
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Assign limit▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、リミット機能に割り当てるプロセス変数を選択します。

選択項目

▪ Off
▪ Concentration▪ Dew Point 1▪ Dew Point 2

初期設定

Concentration

Switch-on value
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Switch-on value▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチオンポイントの測定値を入力します。

選択項目

符号付き浮動小数点数

初期設定

0 ppmv

追加情報

説明
この機能を使用して、スイッチオンの値のリミット値を入力します（プロセス変数 > スイッ
チオンの値 = クローズ、導通）。
ヒステリシスを使用する場合：Switch-on value > Switch-off value

依存関係
単位については、Assign limit パラメータ（→
64139）で選択したプロセス変数に従います。
Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Switch-off value
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Switch-off value▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチオフポイントの測定値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

0 ppmv

追加情報

説明
この機能を使用して、スイッチオフの値のリミット値を入力します（プロセス変数 < スイッ
チオフの値 = オープン、非導通）。
ヒステリシスを使用する場合：Switch-on value > Switch-off value

依存関係
単位については、Assign limit パラメータ（→

139）で選択したプロセス変数に従います。

Assign status
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Assign status▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →で、Status が選択されていること。

説明

この機能を使用して、スイッチ出力の機器ステータスを選択します。

選択項目

▪ Off
▪ Validation Control

初期設定

Off

Switch-on delay
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Switch-on delay▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチ出力のスイッチオンの遅延時間を入力

ユーザー入力0.0～100.0 s

初期設定

0.0 s

Switch-off delay
ナビゲーション
必須条件

Expert → Output → Switch output 1～n → Switch-off delay▪ Operating mode パラメータ →で、Switch が選択されていること。

▪ Switch output function パラメータ →説明
Endress+Hauser

で、Limit が選択されていること。

この機能を使用して、スイッチ出力のスイッチオフの遅延時間を入力
65

機能説明書

J22 TDLAS ガスアナライザ

ユーザー入力0.0～100.0 s

初期設定

0.0 s

Switch state
Expert → Output → Switch output 1～n → Switch state

ナビゲーション
必須条件

Operating mode パラメータ →で、Switch が選択されていること。

説明ステータス出力の現在のステータス切り替えを表示します。

選択項目▪ Open▪ Closed

追加情報

ユーザーインターフェース▪ Open：スイッチ出力は非導通です。
▪ Closed：スイッチ出力は導通です。

Invert output signal
Expert → Output → Switch output 1～n → Invert output signal

ナビゲーション
説明

この機能を使用して、出力信号を反転させるかどうか選択します。

選択項目

▪ No
▪ Yes

初期設定

No

追加情報

選択項目
No（パッシブ - マイナス）

A0026693

Yes（パッシブ - プラス）

A0026692

66

Endress+Hauser

J22 TDLAS ガスアナライザ

3.5.3

機能説明書

Relay output 1～n

ナビゲーション

Expert → Output → Relay output 1～n‣ Relay output 1～nRelay output function→

67

Assign limit→

68

Assign diag. behavior→

68

Assign status→

68

Switch-off value→

68

Switch-off delay→

68

Switch-on value→

70

Switch-on delay→

70

Switch state→70

Powerless relay status→

70

Relay output function
ナビゲーション

Expert → Output → Relay output 1～n → Relay output function

説明

この機能を使用して、リレー出力の出力機能を選択します。

ユーザーインター▪ Closedフェース▪ Open▪ Diagnostic behavior▪ Limit▪ Status

初期設定

Closed

追加情報

選択項目▪ Closed：リレー出力が恒久的にオンになります（クローズ、導通）。
▪ Open：リレー出力が恒久的にオフになります（オープン、非導通）。
▪ Diagnostic behavior：診断イベントの有無を示します。これは診断情報を出力し、システ
ムレベルで適切に反応するために使用されます。
▪ Limit：プロセス変数が規定されたリミット値に達したことを示します。これはプロセスに関する診断情報を出力し、システムレベルで適切に反応するために使用されます。
▪ Status：検証制御が選択された場合に機器ステータスを表示します。

Endress+Hauser67

機能説明書

J22 TDLAS ガスアナライザ

Assign limit
Expert → Output → Relay output 1～n → Assign limit

ナビゲーション
必須条件

Relay output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、リミット値機能に割り当てるプロセス変数を選択します。

選択項目

▪ Off
▪ Concentration▪ Dew Point 1▪ Dew Point 2

初期設定

Off

Assign diagnostic behavior
Expert → Output → Relay output 1～n → Assign diagnostic behavior

ナビゲーション
必須条件

Relay output function パラメータ →で、Diagnostic behavior が選択されていること。

説明

この機能を使用して、リレー出力で表示される診断イベントのカテゴリを選択します。

選択項目▪ Alarm▪ Alarm or warning▪ Warning

初期設定

Alarm

追加情報

説明
診断イベントが未処理の場合、リレー出力はクローズ/導通となります。

選択項目▪ Alarm：リレー出力はアラームカテゴリの診断イベントのみを通知します。
▪ Alarm or warning：リレー出力はアラームおよび警告カテゴリの診断イベントを通知します。
▪ Warning：リレー出力は警告カテゴリの診断イベントのみを通知します。

Assign status
Expert → Output → Relay output 1～n → Assign status

ナビゲーション
必須条件

Relay output function パラメータ →で、Digital Output が選択されていること。

説明

この機能を使用して、リレー出力の機器ステータスを選択します。

選択項目

▪ Off
▪ Validation Control

初期設定

68

Off

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Switch-off value
Expert → Output → Relay output 1～n → Switch-off value

ナビゲーション
必須条件

Relay output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチオフポイントの測定値を入力します。

ユーザー入力

符号付き浮動小数点数

初期設定

0 ppmv

追加情報

説明
この機能を使用して、スイッチオフの値のリミット値を入力します（プロセス変数 < スイッ
チオフの値 = オープン、非導通）。
ヒステリシスを使用する場合：Switch-on value > Switch-off value

依存関係
単位については、Assign limit パラメータ（→

146）で選択したプロセス変数に従います。

Switch-off delay
Expert → Output → Relay output 1～n → Switch-off delay

ナビゲーション
必須条件

Relay output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチ出力のスイッチオフの遅延時間を入力します。

選択項目

0.0～100.0 s

初期設定

0.0 s

Switch-on value
Expert → Output → Relay output 1～n → Switch-on value

ナビゲーション
必須条件

Relay output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチオンポイントの測定値を入力します。

ユーザー入力

符号付き浮動小数点数

選択項目

▪ Off
▪ Validation Control

追加情報

説明
この機能を使用して、スイッチオンの値のリミット値を入力します（プロセス変数 > スイッ
チオンの値 = クローズ、導通）。
ヒステリシスを使用する場合：Switch-on value > Switch-off value

依存関係
単位については、Assign limit パラメータ（→

Endress+Hauser146）で選択したプロセス変数に従います。

69

機能説明書

J22 TDLAS ガスアナライザ

Switch-on delay
Expert → Output → Relay output 1～n → Switch-on delay (0814–1～n)

ナビゲーション
必須条件

Relay output function パラメータ →で、Limit が選択されていること。

説明

この機能を使用して、スイッチ出力のスイッチオンの遅延時間を入力

ユーザー入力0.0～100.0 s

初期設定

0.0 s

Switch state
Expert → Output → Relay output 1～n → Switch state

ナビゲーション
説明

リレー出力の現在のステータスを表示します。

ユーザーインター▪ Openフェース▪ Closed

追加情報

ユーザーインターフェース▪ Open：リレー出力は非導通です。
▪ Closed：リレー出力は導通です。

Powerless relay status
ナビゲーション

Expert → Output → Relay output 1～n → Powerless relay

説明

この機能を使用して、リレー出力の停止状態を選択します。

選択項目▪ Open▪ Closed

初期設定

Open

追加情報

選択項目
▪ Open：リレー出力は非導通です。
▪ Closed：リレー出力は導通です。

70

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

3.6 Communication
ナビゲーション

Expert → Communication‣ Communication

3.6.1

→

71

‣ Modbus configuration→

71‣ Modbus information→

76‣ Modbus data map→

77‣ Web server→

77

Bus address→

71

Baudrate→

72

Data transfer mode→

72

Parity→72

Byte order→73

Telegram delay→

74

Priority IP address→

74

Inactivity timeout→

74

Max connections→

74

Failure mode→75

Bus termination→

75

Fieldbus writing access→

75

Modbus configuration

ナビゲーション Expert → Communication → Modbus configuration‣ Modbus configuration

Bus address
ナビゲーション
必須条件

Endress+Hauser

Expert → Communication → Modbus configuration → Bus address
Modbus RS485 機器

71

機能説明書

J22 TDLAS ガスアナライザ

説明

この機能を使用して、機器アドレスを入力します。

ユーザー入力

1～247

初期設定

247

Baudrate
ナビゲーション

Expert → Communication → Modbus configuration → Baudrate

必須条件

Modbus RS485 機器

説明

この機能を使用して、伝送速度を選択します。

ユーザー入力▪ 1200 BAUD
▪ 2400 BAUD
▪ 4800 BAUD
▪ 9600 BAUD
▪ 19200 BAUD
▪ 38400 BAUD
▪ 57600 BAUD
▪ 115200 BAUD

初期設定

19200 BAUD

Data transfer mode
ナビゲーション

Expert → Communication → Modbus configuration → Data transfer mode

必須条件

Modbus RS485 機器

説明

この機能を使用して、データ伝送モードを選択します。

選択項目▪ ASCII
▪ RTU

初期設定

RTU

追加情報

選択項目
▪ ASCII：読取可能な ASCII 文字の形式でデータを伝送。LRC を介したエラー保護。
▪ RTU：バイナリ形式でデータを伝送。CRC16 を介したエラー保護。

Parityナビゲーション

Expert → Communication → Modbus configuration → Parity

必須条件

Modbus RS485 機器

説明

この機能を使用して、パリティビットを選択します。

選択項目

▪ Odd
▪ Even▪ None / 1 stop bit▪ None / 2 stop bits

72

Endress+Hauser

J22 TDLAS ガスアナライザ

初期設定

Even

追加情報

選択項目

機能説明書

ASCII 選択リストの選択項目：▪ 0 = Even
▪ 1 = Odd
RTU 選択リストの選択項目：▪ 0 = Even
▪ 1 = Odd
▪ 2 = None / 1 stop bit▪ 3 = None / 2 stop bits
Byte order
ナビゲーション

Expert → Communication → Modbus configuration → Byte order

説明

この機能を使用して、送信されるバイトの順序を選択します。伝送順序は、Modbus マスタと調整する必要があります。

選択項目▪ 0-1-2-3▪ 3-2-1-0▪ 1-0-3-2▪ 2-3-0-1

初期設定

1-0-3-2

追加情報説明
Modbus プロトコルによるバイト順序の規格化はありません。ただし、ホストシステムと機器が同じバイト順序を使用しないと、正しいデータ交換ができません。
ホストシステムでバイト順序を変更するには、多くの場合、広範な知識と多大なプログラミ
ング作業が必要となります。このため、Endress+Hauser では Byte order パラメータ → を導入しました。
これにより、ホストシステムの標準設定を使用して、バイト順序を試行錯誤しながら機器で変更することが可能になります。バイト順序を変更しても正しいデータ交換が実現できない
場合は、これに応じてホストシステムのバイト順序の設定を調整しなければなりません。

バイト伝送順序
バイトのアドレス指定、つまり、バイトの伝送順序は、Modbus 仕様には規定されていません。そのため、設定中にマスタとスレーブの間でアドレス指定方法を調整または一致させることが重要です。Byte order パラメータ → を使用すると、機器でこれを設定することができます。
Byte order パラメータ →の選択項目に従って、バイトが伝送されます。

浮動小数
順序
選択項目

1.

2.

1 – 0 – 3 – 2 バイト 1
バイト 0*（MMMMMMMM）（MMMMMMMM）
0 – 1 – 2 – 3 バイト 0
バイト 1（MMMMMMMM）（MMMMMMMM）

Endress+Hauser3.

4.

バイト 3（SEEEEEEE）バイト 2（EMMMMMMM）
バイト 2
バイト 3（SEEEEEEE）（EMMMMMMM）

73

機能説明書

J22 TDLAS ガスアナライザ

2 – 3 – 0 – 1 バイト 2
バイト 3（SEEEEEEE）バイト 0（EMMMMMMM）（MMMMMMM）
3 – 2 – 1 – 0 バイト 3
バイト 2
バイト 1
（SEEEEEEE）（EMMMMMMM） （MMMMMMMM）* = 初期設定、S = 符号、E = 指数、M = 仮数

バイト 1
（MMMMMMM）
バイト 0（MMMMMMMM）

整数
順序
選択項目

1.

2.

1–0–3–2*
バイト 1（MSB）
3–2–1–0
0–1–2–3
バイト 0（LSB）
2–3–0–1* = 初期設定、MSB = 最上位バイト、LSB = 最下位バイト

バイト 0（LSB）
バイト 1（MSB）

文字列
データ長 18 バイトの機器パラメータの例を表示
順序
選択項目

1.

2.

...

1–0–3–2*
バイト 17
バイト 16...
3–2–1–0（MSB）
0–1–2–3
バイト 16
バイト 17...
2–3–0–1（MSB）* = 初期設定、MSB = 最上位バイト、LSB = 最下位バイト

17.

18.

バイト 1

バイト 0
（LSB）
バイト 1

バイト 0
（LSB）

Telegram delay
Expert → Communication → Modbus configuration → Telegram delay

ナビゲーション
必須条件

Modbus RS485 機器

説明

この機能を使用して、機器が Modbus マスタのリクエストテレグラムに応答するまでの遅延時間を入力します。これにより、通信を低速の Modbus RS485 マスタに適合させることが可能です。

ユーザー入力0～100 ms

初期設定

6 ms

Priority IP address
ナビゲーション

Expert → Communication → Modbus configuration → Priority IP address

必須条件

Modbus RS485 機器

説明

サーバー（アナライザ）への接続が保証されるクライアントの IP アドレス

ユーザー入力

符号付き浮動小数点数

初期設定

0.0.0.074

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Inactivity timeout
ナビゲーション

Expert → Communication → Modbus configuration → Inactivity timeout

必須条件

Modbus RS485 機器

説明

Priority IP address 以外の IP アドレスの場合に、クライアントの接続が切れるまでの非アクティブ状態の継続時間

ユーザー入力0～99 s

初期設定

0s

Max connections
ナビゲーション

Expert → Communication → Modbus configuration → Max connections

必須条件

Modbus TCP 機器

説明

Modbus サーバーへの接続数

ユーザー入力

1～4

初期設定

4

Failure mode
ナビゲーション

Expert → Communication → Modbus configuration → Failure mode

説明

この機能を使用して、Modbus 通信を介して診断メッセージが発生した場合に出力される測定値を選択します。

選択項目▪ NaN value▪ Last valid value

初期設定

NaN value

追加情報

選択項目

1

1

▪ NaN value：機器は NaN 値を出力します 。
▪ Last valid value：機器はエラーが発生する前に、最後に有効であった測定値を出力します。このパラメータの影響は、Assign diagnostic behavior パラメータで選択した項目に応じて異なります。
Bus termination
ナビゲーション

Expert → Communication → Modbus configuration → Bus termination

必須条件

Modbus RS485 機器

説明

終端抵抗の有効化または無効化が表示されます。

ユーザーインター
フェース

▪ Off
▪ On

1

非数

Endress+Hauser75

機能説明書

J22 TDLAS ガスアナライザ

初期設定

Off

追加情報

選択項目
▪ Off：終端抵抗が無効になります。
▪ On：終端抵抗が有効になります。
終端抵抗の有効化に関する詳細については、機器の取扱説明書 →の「終端抵抗の有効化」セクションを参照してください。

Fieldbus writing access
ナビゲーション

Expert → Communication → Modbus configuration → Fieldbus writing access

説明

この機能を使用して、フィールドバス（Modbus プロトコル）を介した機器へのアクセスを制限します。

選択項目▪ Read + write▪ Read only

初期設定

Read + write

追加情報

説明
読み取り/書き込み保護が有効な場合、パラメータの制御およびリセットは現場操作でのみ実行できます。操作ツールを介したアクセスはできません。これにより、上位システムへの周期的な測定値伝送が影響を受けることはなく、常に保証されます。

選択項目▪ Read + write：パラメータは、読み取りおよび書き込みパラメータです。
▪ Read only：パラメータは、読み取り専用パラメータです。

3.6.2

Modbus information

ナビゲーション

Expert → Communication → Modbus information‣ Modbus information
Device ID→

76

Device revision→76

Device ID
ナビゲーション
説明

Expert → Communication → Modbus information → Device ID
機器を識別するための機器 ID が表示されます。

ユーザーインターフェース 4 桁の 16 進数
Device revision
ナビゲーション
説明

Expert → Communication → Modbus information → Device revision
機器リビジョンが表示されます。

ユーザーインターフェース 4 桁の 16 進数
76Endress+Hauser

J22 TDLAS ガスアナライザ

3.6.3

機能説明書

Modbus data map

ナビゲーション

Expert → Communication → Modbus data map‣ Modbus data map
Scan list register 0 to 15→

77

Scan list area 0 to 15→

77

Scan list register 0 to 15
ナビゲーション

Expert → Communication → Modbus data map → Scan list register 0 to 15

説明

この機能を使用して、スキャンリストレジスタを入力します。レジスタアドレス（1 ベース）を入力すると、スキャンリストレジスタ 0～15 に機器パラメータを割り当てることによって、最大 16 個の機器パラメータをグループ化できます。ここで割り当てられた機器パ
ラメータのデータは、レジスタアドレス 5051～5081 を介して読み出されます。

ユーザー入力1～65,535

初期設定

1

Scan list area 0 to 15
ナビゲーション

Expert → Communication → Modbus data map → Scan list area 0 to 15

説明

この機能を使用して、スキャンリストエリアを入力します。

ユーザー入力1～65,535

初期設定

1

3.6.4

Web server

ナビゲーション

Expert → Communication → Web server‣ Web serverEndress+Hauser

Web server language→

77

MAC address→

77

DHCP client→77

IP address→

79

Subnet mask→

79

Default gateway→79

Web server functionality

→

79
77

機能説明書

J22 TDLAS ガスアナライザ

Login page→79

Web server language
ナビゲーション

Expert → Communication → Web server → Webserv.language

説明

この機能を使用して、Web サーバーの言語設定を選択します。

ユーザー入力▪ English▪ Français▪ Italiano▪ русский язык (Russian)▪ 中文 (Chinese)

初期設定

English

MAC address
ナビゲーション

Expert → Communication → Web server → MAC Address

説明

機器の MAC アドレスを表示

ユーザー入力英字と数字から成る一意的な 12 桁の文字列

初期設定

各機器に個別のアドレスが付与されます。

追加情報例
表示形式 00:07:05:10:01:5F

DHCP client
Expert → Communication → Web server → DHCP client

ナビゲーション
説明

この機能を使用して、DHCP クライアントの機能を有効/無効にします。

選択項目

▪ Off
▪ On

初期設定

Off

追加情報

影響
Web サーバーの DHCP クライアント機能を選択すると、IP address →、Default gateway →、Subnet mask →が自動的に設定されます。

 機器の MAC アドレスを介して識別されます。
 DHCP client パラメータ → がオンの場合、IP address パラメータ → で設定された IP
アドレス → は無視されます。これは、特に、DHCP サーバーにアクセスできない場合にも当てはまります。IP address パラメータで設定された IP アドレス → は、DHCP
client パラメータ → がオフの場合にのみ使用されます。

78

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

IP address
ナビゲーション

Expert → Communication → Web server → IP address

説明

機器に組み込まれた Web サーバーの IP アドレスを表示または入力します。

ユーザー入力4 オクテット：0～255（特定のオクテットにおいて）初期設定

192.168.1.212

Subnet mask
ナビゲーション

Expert → Communication → Web server → Subnet mask

説明サブネットマスクを表示または入力します。

ユーザー入力4 オクテット：0～255（特定のオクテットにおいて）

初期設定

255.255.255.0

Default gateway
ナビゲーション

Expert → Communication → Web server → Default gateway

説明

デフォルトゲートウェイを表示または入力します。

ユーザー入力4 オクテット：0～255（特定のオクテットにおいて）

初期設定

0.0.0.0

Endress+Hauser

79

機能説明書

J22 TDLAS ガスアナライザ

Web server functionality
Expert → Communication → Web server → Webserver functionality

ナビゲーション
説明

この機能を使用して、Web サーバーのオン/オフを切り替えます。

選択項目

▪ Off
▪ HTML Off
▪ On

初期設定

On

追加情報

説明
▪ Off：Web サーバーは完全に無効になります。
▪ ポート 80 はロックされます。
▪ HTML Off：Web サーバーの HTML バージョンは使用できません。
▪ On：すべての Web サーバー機能が使用できます。
▪ JavaScript が使用されます。
▪ パスワードは暗号化された状態で伝送されます。
▪ パスワードの変更も暗号化された状態で伝送されます。

Login page
ナビゲーション

Expert → Communication → Web server → Login page

説明

この機能を使用して、ログイン画面の形式を選択します。

選択項目▪ Without header▪ With header

初期設定

80

With header

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

3.7 Diagnostics
ナビゲーション

Expert → Diagnostics‣ Diagnostics
Actual diagnostics→

81

Previous diagnostics→

82

Operating time from restart→82

Operating time→82

‣ Diagnostic list→

83

‣ Event logbook→

86‣ Device information→

87‣ Main electronic module + I/O
module 1→89‣ Sensor electronic module (ISEM)→

90‣ I/O module 2→

91‣ I/O module 3→

92‣ Display module→

93‣ Data logging→

93‣ Heartbeat Technology→

94‣ Simulation→

110‣ Spectrum plots→

114‣ SD card→

119

Actual diagnostics
ナビゲーション

Expert → Diagnostics → Actual diagnostics

必須条件

診断イベントが発生していること。

説明

現在の診断メッセージを表示します。2 つあるいはそれ以上のメッセージが同時に発生した
場合は、最優先に処理する必要のあるメッセージが表示されます。

ユーザーインター
フェース

診断動作のシンボル、診断コード、ショートメッセージ

Endress+Hauser81

機能説明書追加情報

J22 TDLAS ガスアナライザ

表示
その他の未処理の診断メッセージは、Diagnostic list サブメニュー → に表示されます。
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、 キーを使用してアクセスできます。

例
表示形式の場合：
F271 Main electronics failure
Previous diagnostics service ID
Expert → Diagnostics → Previous diagnostics

ナビゲーション
必須条件

すでに 2 つの診断イベントが発生していること。

説明

現在のメッセージの前に発生した診断メッセージが表示されます。

ユーザーインター
フェース

0～65,535

追加情報表示
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、キーを使用してアクセスできます。

例
表示形式の場合：
F271 Main electronics failure

Operating time from restart
ナビゲーション

Expert → Diagnostics → Operating time from restart

説明

この機能を使用して、前回、機器を再起動してからの稼働時間を表示します。

ユーザーインター

日（d）、時（h）、分（m）、秒（s）

フェース

Operating time
ナビゲーション

Expert → Diagnostics → Operating time

説明

この機能を使用して、機器の稼働時間を表示します。

ユーザーインター

日（d）、時（h）、分（m）、秒（s）

フェース
追加情報

ユーザーインターフェース
最大日数は 9999 です。これは、27 年に相当します。

82

Endress+Hauser

J22 TDLAS ガスアナライザ

3.7.1

機能説明書

Diagnostic list

ナビゲーション

Expert → Diagnostics → Diagnostic list
‣ Diagnostic list
Diagnostics 1→

83

Diagnostics 2→

83

Diagnostics 3→

84

Diagnostics 4→85

Diagnostics 5→85

Diagnostics 1
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Diagnostics 1
最も優先度の高い現在の診断メッセージを表示します。

ユーザーインターフェース 0～65,535
追加情報表示
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、 キーを使用してアクセスできます。

例
表示形式の場合：
F 271 Main electronics failure
F276 I/O module failure
Timestamp 1
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Timestamp
最も優先度の高い診断メッセージが発生したときの稼働時間を表示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
追加情報表示
診断メッセージは、Diagnostics 1 パラメータ →で確認できます。

例
表示形式の場合：
24d12h13m00s
Diagnostics 2
ナビゲーション
説明
Endress+Hauser

Expert → Diagnostics → Diagnostic list → Diagnostics 2
2 番目に優先度の高い現在の診断メッセージを表示します。
83機能説明書

J22 TDLAS ガスアナライザ

ユーザーインターフェース 0～65,535
追加情報表示
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、 キーを使用してアクセスできます。

例
表示形式の場合：
F271 Main electronics failure
F276 I/O module failure
Timestamp 2
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Timestamp
2 番目に優先度の高い診断メッセージが発生したときの稼働時間を表示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
追加情報表示
診断メッセージは、Diagnostics 2 パラメータ →で確認できます。

例
表示形式の場合：
24d12h13m00s
Diagnostics 3
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Diagnostics 3
3 番目に優先度の高い現在の診断メッセージを表示します。

ユーザーインターフェース 0～65,535
追加情報表示
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、 キーを使用してアクセスできます。

例
表示形式の場合：
F271 Main electronics failure
F276 I/O module failure
Timestamp 3
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Timestamp
3 番目に優先度の高い診断メッセージが発生したときの稼働時間を表示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
追加情報表示
診断メッセージは、Diagnostics 3 パラメータ →84

で確認できます。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

例
表示形式：
24d12h13m00s
Diagnostics 4
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Diagnostics 44 番目に優先度の高い現在の診断メッセージを表示します。

ユーザーインターフェース 0～65,535
追加情報表示
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、 キーを使用してアクセスできます。

例
表示形式の場合：
F271 Main electronics failure
F276 I/O module failure
Timestamp 4
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Timestamp
4 番目に優先度の高い診断メッセージが発生したときの稼働時間を表示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
追加情報表示
診断メッセージは、Diagnostics 4 パラメータ →で確認できます。

例
表示形式の場合：
24d12h13m00s
Diagnostics 5
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Diagnostics 55 番目に優先度の高い現在の診断メッセージを表示します。

ユーザーインターフェース 診断動作のシンボル、診断コード、ショートメッセージ
追加情報表示
現場表示器を使用する場合：診断メッセージの原因に関するタイムスタンプと是正措置には、 キーを使用してアクセスできます。

例
表示形式の場合：
F271 Main electronics failure
F276 I/O module failure

Endress+Hauser85

機能説明書

J22 TDLAS ガスアナライザ

Timestamp 5
ナビゲーション
説明

Expert → Diagnostics → Diagnostic list → Timestamp
5 番目に優先度の高い診断メッセージが発生したときの稼働時間を表示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）

表示

追加情報

診断メッセージは、Diagnostics 5 パラメータ →で確認できます。

例
表示形式の場合：
24d12h13m00s

3.7.2

Event logbook

イベントメッセージの表示
イベントメッセージは時系列で表示されます。イベント履歴には、診断イベントと診断イベントの両方が含まれます。タイムスタンプの前にあるシンボルは、イベントの開始または終了を示します。

ナビゲーション

Expert → Diagnostics → Event logbook‣ Event logbook
Filter options→

86

Filter options
ナビゲーション
説明

Expert → Diagnostics → Event logbook → Filter options
この機能を使用して、現場表示器のイベントログに表示するイベントメッセージのカテゴリ
を選択します。

選択項目

▪ All
▪ Failure (F)▪ Function check (C)▪ Out of specification (S)▪ Maintenance required (M)▪ Information (I)

初期設定

All

追加情報

説明
ステータス信号は VDI/VDE 2650 および NAMUR 推奨 NE 107 に準拠して分類されます。
F = 故障C = 機能チェック
S = 仕様範囲外
M = 要メンテナンス
I = 情報86

Endress+Hauser

J22 TDLAS ガスアナライザ

3.7.3

機能説明書

Device information

ナビゲーション

Expert → Diagnostics → Device info
‣ Device information
Device tag→

87

Serial number→

87

Firmware version→

87

Device name→88

Order code→

88

Extended order code 1→

88

Extended order code 2→

88

Extended order code 3→

88

ENP version→

89

Device tag
ナビゲーション
説明

Expert → Diagnostics → Device information → Device tag
測定点の一意の名称を表示します。この名称によりユーザーの施設内で測定点を迅速に識別できます。これはヘッダーに表示されます。

ユーザーインターフェース 最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）
初期設定

H2O Analyzer

追加情報

ユーザーインターフェース
1

XXXXXXXXX

A0029422

1 表示部のヘッダーテキストの位置表示される文字数は使用される文字に応じて異なります。
Serial number
ナビゲーション
説明

Expert → Diagnostics → Device information → Serial number
機器のシリアル番号を表示
この番号はアナライザの銘板に明記されています。

ユーザーインターフェース 最大 11 文字の英字および数字
Endress+Hauser87

機能説明書追加情報

J22 TDLAS ガスアナライザ

説明
シリアル番号の用途：▪ 機器を迅速に識別するため（Endress+Hauser へのお問合せなどに使用します）▪ デバイスビューワーを使用して詳細な機器情報を取得するため：
www.endress.com/deviceviewer

Firmware version
ナビゲーション
説明

Expert → Diagnostics → Device information → Firmware version
インストールされている機器のファームウェアバージョンを表示

ユーザーインターフェース 形式 xx.yy.zz の文字列
追加情報表示
ファームウェアバージョンは、以下にも記載されています。
▪ 取扱説明書の表紙に明記▪ 変換器の銘板に明記

Device name
ナビゲーション
説明

Expert → Diagnostics → Device information → Device name
変換器の名称を表示します。これは変換器の銘板にも明記されています。

ユーザーインターフェース H2O Analyzer
Order code
ナビゲーション
説明

Expert → Diagnostics → Device information → Order code
機器オーダーコードを表示します。

ユーザーインターフェース 英字、数字、特定の句読記号（/ など）で構成される文字列
追加情報

説明
オーダーコードはセンサおよび変換器の銘板の「オーダーコード」欄に明記されています。
オーダーコードは可逆的な変換プロセスにより拡張オーダーコードから生成されます。拡張
オーダーコードは製品構成に関するすべての機器仕様項目を示すものです。機器仕様項目
を、直接オーダーコードから読み取ることはできません。
オーダーコードの用途：▪ 予備品として同じ機器を注文するため▪ 機器を迅速に識別するため（Endress+Hauser へのお問合せなどに使用します）

Extended order code 1
ナビゲーション
説明88

Expert → Diagnostics → Device information → Extended order code 1
拡張オーダーコードの第 1 部分を表示します。文字数制限があるため、拡張オーダーコード
は最大 3 つに分割されます。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ユーザーインターフェース 文字列

説明

追加情報

拡張オーダーコードは、機器の製品構成に関するすべての仕様項目を示すものであり、それにより機器を一意的に識別することが可能です。
Extended order code 2
ナビゲーション
説明

Expert → Diagnostics → Device information → Extended order code 2
拡張オーダーコードの第 2 部分を表示します。

ユーザーインターフェース 文字列
追加情報

追加情報については、Extended order code 1 パラメータ →を参照してください。

Extended order code 3
ナビゲーション
説明

Expert → Diagnostics → Device information → Extended order code 3
拡張オーダーコードの第 3 部分を表示します。

ユーザーインターフェース 文字列
追加情報

追加情報については、Extended order code 1 パラメータ →を参照してください。

ENP version
ナビゲーション
説明

Expert → Diagnostics → Device information → ENP version
電子銘板のバージョンを表示します。

ユーザーインターフェース 文字列
初期設定

2.02.00

追加情報

説明
この電子銘板には機器 ID のデータ記録が保存され、機器の外側に貼付された銘板より多くのデータが含まれています。

3.7.4

Main electronic module + I/O module 1

ナビゲーション

Expert → Diagnostics → Main electronic +I/O module 1‣ Main electronic module + I/O module 1

Endress+Hauser

Firmware version→

89

Build no. software→

89

Bootloader revision

→

90

89

機能説明書

J22 TDLAS ガスアナライザ

Firmware version
ナビゲーション

Expert → Diagnostics → Main electronic +I/O module 1 → Firmware version
この機能を使用して、モジュールのファームウェアリビジョンを表示します。

説明

ユーザーインターフェース 正の整数
Build no. software
ナビゲーション

Expert → Diagnostics → Main electronic +I/O module 1 → Build no. software
この機能を使用して、モジュールのソフトウェアのビルド番号を表示します。

説明

ユーザーインターフェース 正の整数
Bootloader revision
ナビゲーション

Expert → Diagnostics → Main electronic +I/O module 1 → Bootloader revision
この機能を使用して、ソフトウェアのブートローダリビジョンを表示します。

説明

ユーザーインターフェース 正の整数

3.7.5

Sensor electronic module (ISEM)

ナビゲーション

Expert → Diagnostics → Sens. Electronic‣ Sensor electronic module (ISEM)
Firmware version→

90

Build no. software→

90

Bootloader revision→90

Firmware version
ナビゲーション
説明

Expert → Diagnostics → Sensor electronic module (ISEM) → Firmware version
この機能を使用して、モジュールのファームウェアリビジョンを表示します。

ユーザーインターフェース 正の整数
Build no. software
ナビゲーション
説明

Expert → Diagnostics → Sensor electronic module (ISEM) → Build no. software
この機能を使用して、モジュールのソフトウェアのビルド番号を表示します。

ユーザーインターフェース 正の整数
90

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Bootloader revision
Expert → Diagnostics → Sensor electronic module (ISEM) → Bootloader rev.

ナビゲーション

この機能を使用して、ソフトウェアのブートローダリビジョンを表示します。

説明

ユーザーインターフェース 正の整数

3.7.6

I/O module 2

ナビゲーション

Expert → Diagnostics → I/O module 2‣ I/O module 2
I/O module 2 terminal numbers
Firmware version
Build no. software
Bootloader revision

→

91

→

91

→

91

→

91

I/O module 2 terminal numbers
ナビゲーション
説明

Expert → Diagnostics → I/O module 2 → I/O 2 terminals
I/O モジュールが使用している端子番号を表示します。

ユーザーインターフェース ▪ Not used
▪ 26-27（I/O 1）▪ 24-25（I/O 2）▪ 22-23（I/O 3）
Firmware version
ナビゲーション
説明

Expert → Diagnostics → I/O module 2 → Firmware version
この機能を使用して、モジュールのファームウェアリビジョンを表示します。

ユーザーインターフェース 正の整数
Build no. software
ナビゲーション
説明

Expert → Diagnostics → I/O module 2 → Build no. software
この機能を使用して、モジュールのソフトウェアのビルド番号を表示します。

ユーザーインターフェース 正の整数
Bootloader revision
ナビゲーション
Endress+Hauser

Expert → Diagnostics → I/O module 2 → Bootloader rev.
91

機能説明書説明

J22 TDLAS ガスアナライザ

この機能を使用して、ソフトウェアのブートローダリビジョンを表示します。

ユーザーインターフェース 正の整数

3.7.7

I/O module 3

ナビゲーション

Expert → Diagnostics → I/O module 3‣ I/O module 3
I/O module 3 terminal numbers
Firmware version
Build no. software
Bootloader revision

→

92

→

92

→

92

→

92

I/O module 3 terminal numbers
ナビゲーション
説明

Expert → Diagnostics → I/O module 3 → I/O 3 terminals
I/O モジュールが使用している端子番号を表示します。

ユーザーインターフェース ▪ Not used
▪ 26-27（I/O 1）▪ 24-25（I/O 2）▪ 22-23（I/O 3）
Firmware version
ナビゲーション
説明

Expert → Diagnostics → I/O module 3 → Firmware version
この機能を使用して、モジュールのファームウェアリビジョンを表示します。

ユーザーインターフェース 正の整数
Build no. software
ナビゲーション
説明

Expert → Diagnostics → I/O module 3 → Build no. software
この機能を使用して、モジュールのソフトウェアのビルド番号を表示します。

ユーザーインターフェース 正の整数
Bootloader revision
ナビゲーション
説明

Expert → Diagnostics → I/O module 3 → Bootloader rev.
この機能を使用して、ソフトウェアのブートローダリビジョンを表示します。

ユーザーインターフェース 正の整数
92

Endress+Hauser

J22 TDLAS ガスアナライザ

3.7.8

機能説明書

Display module

ナビゲーション

Expert → Diagnostics → Display module‣ Display module
Firmware version→

93

Build no. software→

93

Bootloader revision→

93

Firmware version
ナビゲーション
説明

Expert → Diagnostics → Display module → Firmware version
この機能を使用して、モジュールのファームウェアリビジョンを表示します。

ユーザーインターフェース 正の整数
Build no. software
ナビゲーション
説明

Expert → Diagnostics → Display module → Build no. software
この機能を使用して、モジュールのソフトウェアのビルド番号を表示します。

ユーザーインターフェース 正の整数
Bootloader revision
ナビゲーション
説明

Expert → Diagnostics → Display module → Bootloader rev.
この機能を使用して、ソフトウェアのブートローダリビジョンを表示します。

ユーザーインターフェース 正の整数

3.7.9

Data logging

ナビゲーション

Expert → Diagnostics → Data logging

 このメニューは、Web サーバー経由でのみ使用できます。アナライザの現場表示器はチャートに対応していません。
‣ Data logging

Endress+Hauser

Assign chan. 1～n→94

Logging interval→

94

Clear logging data→

95

Data logging

→

95
93

機能説明書

J22 TDLAS ガスアナライザ

Logging delay→

95

Data log.control→

96

Data log. status→

96

Logging duration→

97

Assign channel 1～n
ナビゲーション

Expert → Diagnostics → Data logging → Assign channel 1～n

説明

この機能を使用して、データロギングチャンネルのプロセス変数を選択します。

選択項目

▪ Off
▪ Concentration▪ Cell gas pressure▪ Cell gas temperature▪ Dew point 1▪ Dew point 2▪ Current output 1▪ Current output 2▪ Flow switch state

初期設定

Off

追加情報

説明
拡張 HistoROM を使用して、合計 1000 個の測定値をロギングできます。つまり、▪ ロギングチャンネルを 1 つ使用する場合：チャンネルあたりのデータポイント数 1000 個▪ ロギングチャンネルを 2 つ使用する場合：チャンネルあたりのデータポイント数 500 個▪ ロギングチャンネルを 3 つ使用する場合：チャンネルあたりのデータポイント数 333 個▪ ロギングチャンネルを 4 つ使用する場合：チャンネルあたりのデータポイント数 250 個
データポイントが最大数に達すると、データログの最も古いデータポイントが周期的に上書きされ、常に最新の測定値（1000、500、333、または 250 個）がログに保存されます（リングメモリ形式）。
 選択項目の設定を変更すると、ログの内容が削除されます。

Logging interval
ナビゲーション

Expert → Diagnostics → Data logging → Logging interval

説明

この機能を使用して、データロギングの時間間隔 Tlog を入力します。この値は、メモリ内の個々のデータポイント間の時間間隔を決定します。

ユーザー入力0.1～3600.0 s

初期設定

1.0 s

追加情報

説明
これは、データログの各データポイント間の時間間隔を設定するものであり、それによりロギング可能な最大プロセス時間 Tlog が決まります。
▪ ロギングチャンネルを 1 つ使用する場合：Tlog = 1000 × tlog

94

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書▪ ロギングチャンネルを 2 つ使用する場合：Tlog = 500 × tlog
▪ ロギングチャンネルを 3 つ使用する場合：Tlog = 333 × tlog
▪ ロギングチャンネルを 4 つ使用する場合：Tlog = 250 × tlog
この時間が経過すると、データログの最も古いデータポイントが周期的に上書きされ、常に Tlog の時間がメモリに保持されます（リングメモリ形式）。
 ロギングの時間間隔を変更すると、ログの内容が削除されます。

例
ロギングチャンネルを 1 つ使用する場合：▪ Tlog = 1000 × 1 s = 1000 s ≈15 min▪ Tlog = 1000 × 10 s = 10000 s ≈ 3 h▪ Tlog = 1000 × 80 s = 80000 s ≈ 1 d▪ Tlog = 1000 × 3600 s = 3600000 s ≈ 41 d
Clear logging data
ナビゲーション

Diagnostics → Data logging → Clear logging
Expert → Diagnostics → Data logging → Clear logging

説明

この機能を使用して、すべてのログデータを削除します。

選択項目▪ Cancel▪ Clear data

初期設定

Cancel追加情報

選択項目▪ Cancel：データは削除されません。すべてのデータが保存されます。
▪ Clear data：データが削除されます。ロギング処理が最初から開始します。

Data logging
ナビゲーション

Diagnostics → Data logging → Data logging
Expert → Diagnostics → Data logging → Data logging

説明

この機能を使用して、データロギングの方法を選択します。

選択項目▪ Overwriting（上書き）▪ Not overwriting（上書きなし）

初期設定

Overwriting（上書き）

追加情報

選択項目
1

▪ Overwriting：機器メモリには FIFO の原則が適用されます。
▪ Not overwriting：測定値メモリがいっぱいになると、データロギングはキャンセルされます（単発）。

1

FIFO = 先入れ先出しのデータ保存方法

Endress+Hauser95

機能説明書

J22 TDLAS ガスアナライザ

Logging delay
ナビゲーション

Diagnostics → Data logging → Logging delay
Expert → Diagnostics → Data logging → Logging delay

必須条件

Data logging パラメータ →で、Not overwriting が選択されていること。

説明

この機能を使用して、測定値ロギングの遅延時間を入力します。

ユーザー入力0～999 時間

初期設定

0 時間

追加情報

説明
Data logging control パラメータ → を使用してデータロギングを開始すると、入力した遅延時間が経過するまで機器にデータは保存されません。

Data logging control
ナビゲーション

Diagnostics → Data logging → Data logging control
Expert → Diagnostics → Data logging → Data logging control

必須条件

Data logging パラメータ →で、Not overwriting が選択されていること。

説明

この機能を使用して、測定値ロギングを開始または停止します。

選択項目▪ None▪ Delete + start▪ Stop

初期設定

None

追加情報

選択項目
▪ None：最初の測定値ロギングステータスです。
▪ Delete + start：すべてのチャンネルに対して記録された測定値はすべて削除され、測定値
ロギングが再び開始します。
▪ Stop：測定値ロギングが停止します。

Data logging status
ナビゲーション

Diagnostics → Data logging → Data log. status
Expert → Diagnostics → Data logging → Data log. status

必須条件

Data logging パラメータ →で、Not overwriting が選択されていること。

説明

測定値ロギングステータスを表示します。

選択項目▪ Done▪ Delay active▪ Active▪ Stopped

初期設定

Done

追加情報

選択項目
▪ Done：測定値ロギングが実行され、正常に完了しました。
▪ Delay active：測定値ロギングが開始しましたが、ロギングの時間間隔はまだ経過してい
ません。

96

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書▪ Active：ロギングの時間間隔が経過し、測定値ロギングが有効です。
▪ Stopped：測定値ロギングが停止します。
Logging duration
ナビゲーション

Diagnostics → Data logging → Logging duration
Expert → Diagnostics → Data logging → Logging duration

必須条件

Data logging パラメータ →で、Not overwriting が選択されていること。

説明

全ロギング期間を表示します。

選択項目

浮動小数点を含む正の数

初期設定

0s

3.7.10 Heartbeat Technology
Heartbeat 検証 + モニタリングのパラメータの詳細については、機器の個別説明書 →ナビゲーション

7 を参照してください。

Expert → Diagnostics → Heartbeat Technology‣ Heartbeat Technology‣ Heartbeat settings→

97‣ Performing verification→

100‣ Verification results→

105‣ Gas validation results→

108‣ Monitoring results→

110

Plant operator→

97

Location→

97‣ Gas validation settings→

98

Heartbeat settings サブメニュー

ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings‣ Heartbeat settings

Endress+Hauser

97

機能説明書

J22 TDLAS ガスアナライザ

Plant operator
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Plant operator

説明

この機能を使用して、プラントオペレータを入力します。

ユーザー入力最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

Location
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Location

説明

この機能を使用して、場所を入力します。

ユーザー入力最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

Gas validation settings サブメニュー

ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation settings‣ Gas validation settings
Select validation calibration→

98

Validation Type→98

Num Validations→

99

Validation gas purge time→99

Meas. duration→

99

Validation gas information→

99

Validation concentration→

99

Validation allowance→

100

Select validation calibration
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Select validation calibration

説明

検証用の校正を選択します。検証ガスの組成と厳密に適合させる必要があります。

選択項目

▪ 1
▪ 2
▪ 3
▪ 4

初期設定

1

98

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Validation Type
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Validation Type説明

検証ガスの流量が手動（ユーザー制御）または自動（機器制御）のいずれであるかを選択し
ます。

選択項目

▪ Validation manual gas▪ Validation auto gas

初期設定

Validation manual gas

Num Validations
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Num Validations

説明

検証点の数を選択します。

選択項目

1

初期設定

1

Validation gas purge time
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Validation gas purge time

説明

検証ガスのパージ時間を入力します。

ユーザー入力

0～5 分

初期設定

1.00 分

Meas. Duration
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Meas. duration

説明

測定の統計データ（平均、標準偏差）を計算する時間を入力します。

ユーザー入力0.25～60 分

初期設定

1.00 分Validation gas information
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Validation gas information

説明

検証ガスのソースの説明または識別情報を入力します（ガス流、ボトル、ボトルのシリアル
番号）。

ユーザー入力最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

初期設定

Unknown validation gas

Endress+Hauser99

機能説明書

J22 TDLAS ガスアナライザ

Validation concentration
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Validation concentration

説明

検証ガス内の被分析物の濃度を入力します。

ユーザー入力0～1000000 ppmv

初期設定

0 ppmv

追加情報

Validation concentration の値は濃度単位に応じて異なります。

Validation allowance
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation
settings → Validation allowance

説明

検証濃度と測定濃度間の許容偏差を設定します。

ユーザー入力0～100 %

初期設定

0.0000%

Performing verification ウィザード

ナビゲーション

Expert → Diagnostics → Heartbeat Techn. → Perform.verific.
‣ Performing verification

100

Year

→

100

Month

→

101

Day

→

101

Hour

→

101

AM/PM

→

102

Minute→102

Meas. Duration→

102

Verification mode→

102

Ext. device info→

103

Start verification→

103

Progress→

103

Measured val.

→

103

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Output values→

104

Measured conc.

→

104

Status→104

Verification result

→

104

Year
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Year

必須条件

Heartbeat 検証がアクティブでない場合に編集可能。

説明

この機能を使用して、検証を実施した年を入力します。

ユーザー入力9 ～ 99

初期設定

21

Month
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Month

必須条件

Heartbeat 検証がアクティブでない場合に編集可能。

説明

この機能を使用して、検証を実施した月を選択します。

ユーザー入力▪ January▪ February▪ March▪ April▪ May▪ June▪ July▪ August▪ September▪ October▪ November▪ December

初期設定

JanuaryDay
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Day

必須条件

Heartbeat 検証がアクティブでない場合に編集可能。

説明

この機能を使用して、検証を実施した日を入力します。

ユーザー入力1～31 d

初期設定

1d

Endress+Hauser101

機能説明書

J22 TDLAS ガスアナライザ

Hour
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Hour

必須条件

Heartbeat 検証がアクティブでない場合に編集可能。

説明

この機能を使用して、検証を実施した時間を入力します。

ユーザー入力0～23 h

初期設定

12 h

AM/PM
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → AM/PM

必須条件

Heartbeat 検証がアクティブでない場合に編集可能。
Date/time format パラメータ → で、dd.mm.yy hh:mm am/pm または mm/dd/yy hh:mm
am/pm が選択されていること。

説明

この機能を使用して、12 時間表示の場合に午前（AM）または午後（PM）の時間入力を選択します。

ユーザー入力

▪ AM
▪ PM

初期設定

AM

Minuteナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Minute

必須条件

Heartbeat 検証がアクティブでない場合に編集可能。

説明

この機能を使用して、検証を実施した分を入力します。

ユーザー入力0～59 分

初期設定

0分

Meas. duration
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Meas.
Duration

必須条件

検証ステータスがアクティブになっていない場合は、編集が可能です。

説明

測定の統計データ（平均、標準偏差）を計算する時間を入力します。

ユーザー入力0.25～60 分

Verification mode
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Verification
mode

必須条件

検証ステータスがアクティブになっていない場合は、編集が可能です。

102

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

説明

検証モードを選択します。
▪ Standard verification：内部検証は、外部で測定した変数を手動で確認することなく、機器により自動的に実行されます。
▪ Extended validation：Standard verification に似ていますが、検証基準ガスを使用して測定を行います。
▪ Extended current output：Standard verification に似ていますが、検証基準ガスを使用して測定を行います。
▪ Extended validation and current output：Extended validation と Extended current outputの両方を実行します。

選択項目▪ Standard verification▪ Extended validation▪ Extended current output▪ Extended validation and current output

初期設定Standard verification

External device information
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Ext. device
info

必須条件

以下の条件を満たしていること。
Verification mode パラメータ → で、Extended current output または Extended validationand current output が選択されていること。
検証ステータスがアクティブになっていない場合に、編集可能であること。

説明

拡張検証用の機器の記録

ユーザー入力最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）。

Start verification
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Start verification

説明

検証の開始。
完全な検証を実施するには、選択パラメータを個別に選択します。外部測定値の記録の完了
後に、Start を使用して検証を開始できます。

選択項目▪ Cancel1▪ Output 1 low value
1▪ Output 1 high value
1▪ Output 2 low value
1▪ Output 2 high value▪ Start▪ Prepare validation▪ End validation

初期設定

Cancel1

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser

103機能説明書

J22 TDLAS ガスアナライザ

Progress
ナビゲーション
説明

Expert → Diagnostics → Heartbeat Technology → Performing verification → Progress
プロセスの進捗が示されます。

ユーザーインターフェース 0～100 %Measured values
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Measured
values

必須条件

Start verification パラメータ →▪ Output 1 low value▪ Output 1 high value▪ Output 2 low value▪ Output 2 high value

説明

この機能を使用して、外部測定変数の電流出力の測定値（実際値）を入力します。電流出力の単位は [mA] です。

ユーザー入力

符号付き浮動小数点数

初期設定

0

で、以下の選択項目のいずれかが選択されていること。

Output values
ナビゲーション
説明

Expert → Diagnostics → Heartbeat Technology → Performing verification → Output values
外部測定変数の電流出力のシミュレーション出力値（目標値）を表示します。電流出力の単位は [mA] です。

ユーザーインターフェース 符号付き浮動小数点数
Measured concentration
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Measured
concentration

説明

拡張検証中の検証ガスの濃度を表示します。

ユーザーインターフェース 0～1000000 ppmv
Status
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Status

説明

検証の現在のステータスを表示します。

ユーザーインター
フェース▪ Done▪ Busy▪ Failed▪ Not done
▪ Purging104

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Verification result
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Performing verification → Verification
Result説明

検証の全体結果を表示します。

ユーザーインターフェース ▪ Not supported▪ Passed▪ Not done
▪ Failed▪ Not plugged
初期設定

Not done

Verification results サブメニュー

ナビゲーション

Expert → Diagnostics → Heartbeat Techn. → Verific. Results

Verification results
Date/time (manually entered)→

105

Verification ID→

105

Operating time→

106

Verification result→

106

Sensor→106

Sens. electronic→106

Gas validation→

107

I/O module→

107

System status→

107

Date/time (manually entered)
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → Date/time

必須条件

検証が実行されました。

説明

日付と時刻

ユーザーインターフェース dd.mmmm.yyyy; hh:mm
初期設定
Endress+Hauser

1 January 2010; 12:00
105

機能説明書

J22 TDLAS ガスアナライザ

Verification ID
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → Verification ID

必須条件

検証が実行されました。

説明

機器の検証結果の連続番号付けを表示します。

ユーザーインターフェース 0～65,535
初期設定

0Operating time
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → Operating time

必須条件

検証が実行されました。

説明

検証までの機器の稼働時間を示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
Verification result
ナビゲーション
説明

Expert → Diagnostics → Heartbeat Technology → Verification results → Verification result
検証の全体結果を表示します。

ユーザーインターフェース ▪ Not supported▪ Passed▪ Not done
▪ Failed
初期設定

Not done

Sensor
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → Sensor

必須条件

Verification result パラメータ →説明

センサの結果を表示します。

で、Failed の結果が表示されていること。

ユーザーインターフェース ▪ Not supported▪ Passed▪ Not done
▪ Failed
初期設定

Not done

Sensor electronic module (ISEM)
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → Sens. Electronic

必須条件

Verification result パラメータ →説明

センサ電子モジュール（ISEM）の結果を表示します。

106

で、Failed の結果が表示されていること。

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ユーザーインターフェース ▪ Not supported▪ Passed▪ Not done
▪ Failed
初期設定

Not doneGas validation
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → Gas validation

必須条件

Verification result パラメータ →説明

ガス検証の結果を表示します。

で、Failed の結果が表示されていること。

ユーザーインターフェース ▪ Failed▪ Passed▪ Not done
▪ Not supported▪ Not plugged
初期設定

Not doneI/O module
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → I/O module

必須条件

Verification result パラメータ →で、Failed が表示されていること。

説明I/O モジュールの I/O モジュール監視の結果を表示します。
▪ 電流出力：電流の精度▪ 電流入力：電流の精度▪ リレー出力：スイッチング回数
Heartbeat 検証では、デジタル入力/出力はチェックされず、結果は出力されません。

ユーザーインターフェース ▪ Not supported▪ Passed▪ Not done
▪ Not plugged▪ Failed
初期設定

Not done

システムステータス
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Verification results → System status

必須条件

Verification result パラメータ →で、Failed の結果が表示されていること。

説明

システム状態を表示します。アクティブなエラーに対して機器をテストします。

ユーザーインターフェース ▪ Not supported▪ Passed
Endress+Hauser

107機能説明書

J22 TDLAS ガスアナライザ▪ Not done
▪ Failed
初期設定

Not doneGas validation results サブメニュー

ナビゲーション

Expert → Diagnostics → Heartbeat Techn. → Gas validation results

Gas validation results
Date/time (manually entered)→108

Operating time→108

Gas validation→

108

Concentration average→

109

Conc. std. dev.

→

109

Conc. max→

109

Conc. min→109

Date/time (manually entered)
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Date/time

必須条件

検証が実行されました。

説明

日付と時刻

ユーザーインターフェース dd.mm.yy hh:mm（選択した日時フォーマットに応じて異なります）
初期設定

1 January 2010; 12:00

Operating time
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Operating time

必須条件

検証が実行されました。

説明

検証までの機器の稼働時間を示します。

ユーザーインターフェース 日（d）、時（h）、分（m）、秒（s）
Gas validation
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Gas validation

必須条件

検証が実行されました。

説明

ガス検証完了後のステータス

108

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ユーザーインターフェース ▪ Not supported▪ Passed▪ Not done
▪ Not plugged▪ Failed
Concentration average
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Concentration
average

必須条件

検証が実行されました。

説明

0～1000000 ppmv

ユーザーインターフェース 検証時に測定された平均ガス濃度
Concentration standard deviation
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Conc. Std. dev.

必須条件

検証が実行されました。

説明

検証時に測定された濃度標準偏差の正の浮動小数点値

ユーザーインターフェース 0～1000000 ppmv
Concentration maximum
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Conc. Max

必須条件

検証が実行されました。

説明

ガス検証時に測定された最大濃度

ユーザーインターフェース 0～1000000 ppmv
Concentration minimum
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Gas validation results → Conc. Min

必須条件

検証が実行されました。

説明

ガス検証時に測定された最小濃度

ユーザーインターフェース 0～1000000 ppmv

Endress+Hauser

109

機能説明書

J22 TDLAS ガスアナライザ

Monitoring results サブメニュー

ナビゲーション

Expert → Diagnostics → Heartbeat Techn. → Monitor. results
Monitoring results
Detector reference level→

110

Peak 1 index delta→

110

Peak 2 index delta→

110

Detector reference level
ナビゲーション

Expert → Diagnostics → Heartbeat Technology → Monitor. results → Detector reference
level

説明

光学式検出器からの信号

ユーザーインターフェース 0～5 mA
Peak 1 index delta
ナビゲーション
説明

Expert → Diagnostics → Heartbeat Technology → Monitor. results → Peak 1 index delta
ターゲットピーク 1 値と現在のピーク 1 値の差

ユーザーインターフェース -511.0～+511.0
Peak 2 index delta
ナビゲーション
説明

Expert → Diagnostics → Heartbeat Technology → Monitor. results → Peak 2 index delta
ターゲットピーク 2 値と現在のピーク 2 値の差

ユーザーインターフェース -511.0～+511.0

3.7.11 Simulation
ナビゲーション

Expert → Diagnostics → Simulation‣ Simulation

110

Curr.inp 1～n sim.

→

110

Value curr.inp1～n→110

Curr.outp1～n sim.

→

111

Curr.outpval. 1～n→111

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Switch sim. 1～n→111

Switch state 1～n→

112

Relay out.1～n sim→

112

Switch state 1～n→112

Dev. alarm sim.

→

113

Event category→

113

Diag. event sim.

→

113

Current input 1～n simulation
ナビゲーション

Expert → Diagnostics → Simulation → Current input 1～n sim.

説明

電流入力シミュレーションをオン/オフするためのオプションです。シミュレーション実行
中は、測定値と「機能チェック」カテゴリ（C）の診断メッセージが表示部に交互に示されます。
必要なシミュレーション値は、Value current input 1～n パラメータで設定します。

選択項目

▪ Off
▪ On

初期設定

Off

追加情報

選択項目
▪ Off：電流シミュレーションがオフになります。機器は通常の測定モードになっているか、または別のプロセス変数のシミュレーション中です。
▪ On：電流シミュレーションが作動します。

Value current input 1～n
ナビゲーション

Expert → Diagnostics → Simulation → Value current input 1～n

必須条件

Current input 1～n simulation パラメータで、On が選択されていること。

説明

この機能を使用して、シミュレーション用の電流値を入力します。これにより、電流入力の適切な設定、および上流側のフィードユニットが正しく機能することを確認できます。

ユーザー入力0～22.5 mA

Current output 1～n simulation
ナビゲーション

Expert → Diagnostics → Simulation → Current output 1～n sim.

説明

電流出力のシミュレーションのオン/オフを切り替えます。シミュレーション実行中は、測定値と「機能チェック」カテゴリ（C）の診断メッセージが表示部に交互に示されます。

選択項目

▪ Off
▪ On

初期設定

Off

Endress+Hauser

111

機能説明書追加情報

J22 TDLAS ガスアナライザ

説明
必要なシミュレーション値は、Value current output 1～n パラメータで設定します。

選択項目▪ Off：電流シミュレーションがオフになります。機器は通常の測定モードになっているか、または別のプロセス変数のシミュレーション中です。
▪ On：電流シミュレーションが作動します。
Current output value 1～n
ナビゲーション

Expert → Diagnostics → Simulation → Current output value 1～n

必須条件

Current output 1～n simulation パラメータで、On が選択されていること。

説明

この機能を使用して、シミュレーション用の電流値を入力します。これにより、電流出力が適切に調整されていることを確認できます。

ユーザー入力0～22.5 mA

追加情報

依存関係
入力範囲は、Current span パラメータ →で選択する項目に応じて異なります。

Switch output simulation 1～n
ナビゲーション

Expert → Diagnostics → Simulation → Switch output simulation 1～n

必須条件

Operating mode パラメータ →で、Switch が選択されていること。

説明

この機能を使用して、スイッチ出力のシミュレーションをオン/オフします。シミュレー
ション実行中は、測定値と「機能チェック」カテゴリ（C）の診断メッセージが表示部に交互に示されます。

選択項目

▪ Off
▪ On

初期設定

Off

追加情報

説明
必要なシミュレーション値は、Switch state 1～n パラメータで設定します。

選択項目▪ Off：スイッチシミュレーションがオフになります。機器は通常の測定モードになっているか、または別のプロセス変数のシミュレーション中です。
▪ On：スイッチシミュレーションが有効です。
Switch state 1～n
ナビゲーション

Expert → Diagnostics → Simulation → Switch state 1～n

説明

この機能を使用して、シミュレーション用のスイッチの値を選択します。これにより、ス
イッチ出力の適切な調整、および接続されたスイッチングユニットが正しく機能することを確認できます。

選択項目▪ Open▪ Closed112

Endress+Hauser

J22 TDLAS ガスアナライザ

追加情報

機能説明書

選択項目
▪ Open：スイッチシミュレーションがオフになります。機器は通常の測定モードになっているか、または別のプロセス変数のシミュレーション中です。
▪ Closed：スイッチシミュレーションが有効です。

Relay output 1～n simulation
Expert → Diagnostics → Simulation → Relay out. 1～n sim

ナビゲーション
説明

この機能を使用して、リレー出力のシミュレーションをオン/オフします。シミュレーショ
ン実行中は、測定値と「機能チェック」カテゴリ（C）の診断メッセージが表示部に交互に示されます。

選択項目

▪ Off
▪ On

初期設定

Off

追加情報

説明
必要なシミュレーション値は、Switch state 1～n パラメータで設定します。

選択項目▪ Off：リレーシミュレーションがオフになります。機器は通常の測定モードになっているか、または別のプロセス変数のシミュレーション中です。
▪ On：リレーシミュレーションが作動します。
Switch state 1～n
ナビゲーション

Expert → Diagnostics → Simulation → Switch state 1～n

必須条件

Switch output simulation 1～n パラメータで、On が選択されていること。

説明

この機能を使用して、シミュレーション用のリレー値を選択します。これにより、リレー出力の適切な調整、および接続されたスイッチングユニットが正しく機能することを確認でき
ます。

選択項目

▪ Open▪ Closed

追加情報

選択項目
▪ Open：リレーシミュレーションがオフになります。機器は通常の測定モードになっているか、または別のプロセス変数のシミュレーション中です。
▪ Closed：リレーシミュレーションが作動します。

Device alarm simulation
ナビゲーション

Expert → Diagnostics → Simulation → Device alarm simulation

説明

この機能を使用して、機器アラームをオン/オフします。

選択項目

▪ Off
▪ On

初期設定

Off

Endress+Hauser

113

機能説明書追加情報

J22 TDLAS ガスアナライザ

説明
シミュレーション実行中は、測定値と「機能チェック」カテゴリ（C）の診断メッセージが表示部に交互に示されます。

Diagnostic event category
ナビゲーション

Expert → Diagnostics → Simulation → Event category

説明

この機能を使用して、Diagnostic event simulation パラメータ →示される診断イベントのカテゴリを選択します。

選択項目▪ Sensor▪ Electronics▪ Configuration▪ Process

初期設定

Processのシミュレーションに表Diagnostic event simulation
ナビゲーション

Expert → Diagnostics → Simulation → Diag. event sim.

説明

この機能を使用して、シミュレーション用の診断イベントを選択します。

選択項目

▪ Off
▪ 診断イベント選択リスト（選択したカテゴリに応じて異なる）

初期設定

Off

追加情報

説明
シミュレーションに対して、Diagnostic event category パラメータ →の診断イベントから選択できます。

で選択したカテゴリ

3.7.12 Spectrum plots
ナビゲーション

Expert → Diagnostics → Spectrum plots‣ Spectrum plots

114

Midpoint default 1～n→114

Ramp default 1～n→115

Concentration→

115

Dew point 1→

115

Dew point 2→

115

Cell gas press.

→

116

Cell gas temp.

→

116

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Detect. ref. lvl→

116

Detect. zero lvl→

116

Peak 1 index→

116

Peak1 idx.delta→

117

Peak 2 index→

117

Peak2 idx.delta→

117

Peak track index→

117

Pk trk idx delta→

118

Midpoint delta→

118

Analyzer control

→

118

Reset

→

118

Det. 1 TIA gain→

119

Midpoint default 1～n
ナビゲーション

Expert → Diagnostics → Spectrum plots → Midpoint default 1～n

説明

この値は、ピーク位置を最適化するための中心点デルタの始点として利用できます。

選択項目

0～120 mA

追加情報

工場校正で設定されたピーク中心点値

Ramp default 1～n
ナビゲーション

Expert → Diagnostics → Spectrum plots → Ramp default 1～n

説明

各校正ガスの工場校正されたランプを表示します。

選択項目

0～120 mA

追加情報

レーザーランプはスペクトルのスキャン幅を示します。

Concentration
ナビゲーション

Expert → Diagnostics → Spectrum plots → Concentration

説明

ガス流内の被分析物の測定濃度

選択項目

0～1000000 ppmv

追加情報

被分析物の測定濃度をグラフ化します。

Endress+Hauser

115

機能説明書

J22 TDLAS ガスアナライザ

Dew point 1
ナビゲーション

Expert → Diagnostics → Spectrum plots → Dew point 1

説明

現在計算されている水分露点 1 の温度を表示します。

選択項目

符号付き浮動小数点数

追加情報

単位は、temperature unit パラメータ →の設定が使用されます。露点は所定の濃度と圧力において水分が凝結して液化を開始する温度です。水分の露点計算には当該産業分野で認められた複数の方法があります。詳細については、取扱説明書 BA02152C →

を参照して

ください。

Dew point 2
ナビゲーション

Expert → Diagnostics → Spectrum plots → Dew point 2

説明

現在計算されている水分露点 2 の温度を表示します。

選択項目

符号付き浮動小数点数

追加情報

単位は、temperature unit パラメータ →の設定が使用されます。露点は所定の濃度と圧力において水分が凝結して液化を開始する温度です。水分の露点計算には当該産業分野で認められた複数の方法があります。詳細については、取扱説明書 BA02152C →

を参照して

ください。

Cell gas pressure
ナビゲーション

Expert → Diagnostics → Spectrum plots → Cell gas pressure

説明

サンプルセルで現在測定されているガス圧力をグラフ化します。

選択項目-0.5～+6.9 Bar

追加情報

単位は、pressure unit パラメータ →の設定が使用されます。測定中のサンプルセルの現在の圧力。

Cell gas temperature
ナビゲーション

Expert → Diagnostics → Spectrum plots → Cell gas temperature

説明

サンプルセルで現在測定されているガス圧力を表示します。

選択項目-20～+60 °C

追加情報

単位は、temperature unit パラメータ →の設定が使用されます。測定中のサンプルセルの現在の温度。

116

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Detector reference level
ナビゲーション

Expert → Diagnostics → Spectrum plots → Detector reference level

説明

現在測定されているレーザー検出器の基準レベルをグラフ化します。

選択項目

0～5 mA

追加情報

DC レーザー電流の大きさ。値が範囲外である場合、オプティックの洗浄が必要であること、または配置に問題があることを示している可能性があります。

Detector zero level
ナビゲーション

Expert → Diagnostics → Spectrum plots → Detector zero level

説明

現在測定されているレーザー検出器のゼロレベルを表示します。

選択項目

0～5 mA

追加情報

レーザーがオフのときの DC レーザー出力（例：暗電流）

Peak 1 index
ナビゲーション

Expert → Diagnostics → Spectrum plots → Peak 1 index

説明

現在測定されている 2f スペクトルの吸収ピーク 1 指標位置を表示します。

選択項目0～511.0

追加情報

スキャンに沿った吸収ピークの位置

Peak 1 index delta
ナビゲーション

Expert → Diagnostics → Spectrum plots → Peak 1 index delta

説明

ピーク 1 指標デルタを表示します。

選択項目-511.0～+511.0

追加情報

ピーク 1 指標デルタは、ターゲットピーク 1 値と現在のピーク 1 値の差です。

Peak 2 index
ナビゲーション

Expert → Diagnostics → Spectrum plots → Peak 2 index

説明

現在測定されている 2f スペクトルの吸収ピーク 2 指標位置を表示します。

選択項目0～511.0

追加情報

スキャンに沿った第 2 ピークの位置。ピーク追跡のために使用されます。

Endress+Hauser

117

機能説明書

J22 TDLAS ガスアナライザ

Peak 2 index delta
Expert → Diagnostics → Spectrum plots → Peak 2 index delta

ナビゲーション
説明

ピーク 2 指標デルタを表示します。

選択項目-511.0～+511.0

追加情報

ピーク 2 指標デルタは、ターゲットピーク 2 値と現在のピーク 2 値の差です。

Peak track index
Expert → Diagnostics → Spectrum plots → Peak track index

ナビゲーション
説明

現在測定されている 2f スペクトルのピーク追跡に使用されるピークのピーク追跡指標を表
示します。

選択項目

0～511.0

追加情報

Peak track analyzer control パラメータで Off を選択した場合、この値はゼロになります。それ以外の場合は、この値はピーク追跡に使用されるピークに応じて、Peak 1～n index パラ
メータと同じになります。

Peak track index delta
ナビゲーション

Expert → Diagnostics → Spectrum plots → Peak track index delta

説明

現在測定されている 2f スペクトルのピーク追跡指標とターゲット指標の差を表示します。

選択項目-511.0～+511.0

追加情報

Peak track analyzer control パラメータで Off を選択した場合、この値はゼロになります。それ以外の場合は、この値はピーク追跡に使用されるピークに応じて、Peak 1～n index delta
パラメータと同じになります。

Midpoint delta
ナビゲーション

Expert → Diagnostics → Spectrum plots → Midpoint delta

説明

校正された中心点値と現在使用されている中心点値の差を表示します。

選択項目

0～120 mA

追加情報

Peak track analyzer control パラメータで Off を選択した場合、この値はゼロになります。それ以外の場合は、この値はピーク追跡アルゴリズムによって校正された中心点値に適用される変化量になります。

Analyzer control
ナビゲーション
説明

118

Expert → Diagnostics → Spectrum plots → Analyzer control
ピーク追跡の有効化/無効化を制御します。

Endress+Hauser

J22 TDLAS ガスアナライザ

選択項目

機能説明書

▪ Off
▪ On

初期設定

Off

追加情報

アナライザのピーク追跡のオン/オフを切り替えます。校正ごとに個別のピーク追跡設定があります。通常動作では、ピーク追跡をオンにする必要があります。

ResetExpert → Diagnostics → Spectrum plots → Reset

ナビゲーション
説明

アナライザのピーク中心点の現在値をリセットします。

選択項目

▪ Off
▪ Reset

初期設定

Off

追加情報

リセットにより、アナライザのピーク中心点の現在値が最初の校正後のピーク位置に変更されます。

Det. 1 TIA gain
ナビゲーション

Expert → Diagnostics → Spectrum plots → Det. 1 TIA gain

説明

TIA ゲイン値を表示します。

選択項目

0～15

追加情報

トランスインピーダンスアンプ（TIA）のゲイン値。

3.7.13 SD card
ナビゲーション

Expert → Diagnostics → SD card
‣ SD card
Spectra log rate→

119

Number of spectra files→

119

Validation log level→

120

Number of validation files→

120

Spectra log rate
ナビゲーション
Endress+Hauser

Expert → Diagnostics → SD card → Spectra log rate
119

機能説明書

J22 TDLAS ガスアナライザ

説明

スペクトルデータを SD カードに保存する頻度

選択項目

45～86400 s

初期設定

3600 s

追加情報

通常動作では、1 日に 1 つのスペクトルログファイルが生成されますが、ロギング頻度を高めると、1 日に複数のファイルが生成されます。

Number of spectra files
ナビゲーション
説明

Expert → Diagnostics → SD card → Number of spectra file
推定されるスペクトルファイル数

ユーザーインターフェース 0～30
追加情報

1

アナライザは最大 30 個のスペクトルログファイルに対応します。ファイルは FIFO 方式で保存されます。SD カードの容量が小さい場合は、ファイル数が少なくなります。

Validation log level
ナビゲーション

Expert → Diagnostics → SD card → Validation log level

説明Heartbeat 拡張検証時に検証ログファイルに記録する情報量を指定します。

選択項目

▪ Off
▪ Normal▪ Extended
▪ All

初期設定

Normal

追加情報▪ Off：検証ログ情報は作成されません。
▪ Normal：検証測定中に、トレンド、最初/中間/最後のスペクトル、および検証結果が記録されます。
▪ Extended：検証測定中に、Normal ログレベルに加えてすべてのスペクトルも記録されます。
▪ All：Extended ログレベルに加えて、検証前後のパージ中のすべてのトレンドとスペクトルも記録されます。

Number of validation files
ナビゲーション
説明

Expert → Diagnostics → SD card → Number of validation files
SD カードに保存されている現在の検証ファイル数

ユーザーインターフェース 0～60
追加情報

1

SD カードの容量が 1GB 未満の場合、最大ファイル数は 30 まで減少します。

FIFO = 先入れ先出しのデータ保存方法

120

Endress+Hauser

J22 TDLAS ガスアナライザ

4

機能説明書

認定固有の初期設定

4.1 SI 単位
4.1.1

システム単位プロセス変数

単位

温度

°C

圧力

bar a

4.1.2

フルスケール値

初期設定は以下のパラメータに適用されます。
 20 mA value（電流出力のフルスケール値）
 100% bar graph value 14.1.3

電流出力スパン出力

電流範囲

電流出力 1...n4～20 mA NAMUR

4.2 US 単位
4.2.1

システム単位プロセス変数

単位

温度

°F

圧力

psi a

4.2.2

フルスケール値

初期設定は以下のパラメータに適用されます。
 20 mA value（電流出力のフルスケール値）
 100% bar graph value 14.2.3

電流出力スパン出力

電流範囲

電流出力 1...n4～20 mA US

Endress+Hauser

121

機能説明書5

J22 TDLAS ガスアナライザ

単位の短縮表記の説明

5.1 SI 単位
プロセス変数単位

説明

圧力

Pa a, kPa a, MPa a

パスカル、キロパスカル、メガパスカル
（絶対圧）

bar

バール

Pa g, kPa g, MPa g

パスカル、キロパスカル、メガパスカル（相対/ゲージ圧）

bar g

バール（相対/ゲージ圧）

温度

°C, K

摂氏、ケルビン時間s, m, h, d, y

秒、分、時、日、年

プロセス変数

単位

説明

圧力

psi a

ポンド/平方インチ（絶対圧）psi gポンド/平方インチ（ゲージ圧）

温度°F, °R

華氏、ランキン

時間s, m, h, d, y

秒、分、時、日、年

am, pm

午前、午後

プロセス変数単位説明

時間s, m, h, d, y

秒、分、時、日、年

am, pm

午前、午後

5.2 US 単位

5.3 英国単位

122

Endress+Hauser

J22 TDLAS ガスアナライザ

6

機能説明書

Modbus レジスタ情報

6.1 注記特に記載のない限り、Modbus という表記は Modbus TCP/RS485 機器を表します。

6.1.1

レジスタ情報の構造パラメータ説明は次のセクションに個別に記載されています。
ナビゲーション：パラメータへのナビゲーションパス
パラメータ

レジスタ

データ型

アクセスタイプ

ユーザーインターフェース→/選択項目/ユーザー入力
パラメータ名小数値の形式で表示

浮動小数点数長さ =可能なパラメータへ

選択項目

標準パラメータ説明の4 バイトのアクセスタイプ：

パラメータの個々のページ番号および相互整数長さ = 2 バイト

機能コード 03、オプションのリスト

参照

文字列の長さ、パラメータ04、23 による読み取オプション 1に応じて異なるりアクセス

オプション 2

機能コード 06、16、23 による書き込

オプション 3 （+）（+）= 初期

みアクセス

設定は、国、注文オ
プション、機器設定に応じて異なる
ユーザー入力パラメータの入力値
または入力範囲

MODBUS 機能コード 06、16 または 23 を使用して、不揮発性の機器パラメータを変更した場合、変更内容は機器の EEPROM に保存されます。
 EEPROM への書込み回数は、技術的に最大 100 万回に制限されます。
 この制限を超過した場合、データ損失や機器の故障が発生する可能性があるため、制限を順守してください。
 不揮発性の機器パラメータへの MODBUS を使用した頻繁な書込みは避けてください。

6.1.2

アドレスモデル

機器の Modbus レジスタアドレスは、「Modbus アプリケーションプロトコル仕様 V1.1」に準拠して実装されます。Modbus プロトコルでは、アドレスは 16 ビット（0～65,535 の数値）を使用してエンコードされます。これらは 0 ベースのアドレスです。したがって、Modbus プロトコルアドレスは、レジスタから 1 を引いた値と等しく
なります。
機能コード

アクセスタイプ「Modbus アプリケーションプロトコル仕様」に準拠するレジスタ

03

読み取り

XXXX

04

例：9455 濃度

23
06
16

書き込み

XXXX
例：2439 濃度単位

23

Endress+Hauser

123

機能説明書

J22 TDLAS ガスアナライザ

6.2 エキスパート操作メニューの概要
以下の表は、エキスパート操作メニューとそのパラメータのメニュー構造の概要を示したものです。サブメニュー
またはパラメータの説明については、参照ページをご覧ください。
ExpertLocking status→

10

User role→

11

Enter access code→11

‣ System

→

12

→

12

Display language→

13

Format display→

13

Value 1 display→

140% bargraph 1→

14100% bargraph 1→

14

Decimal places 1→

15

Value 2 display→

16

Decimal places 2→

16

Value 3 display→

170% bargraph 3→

17100% bargraph 3→

17

Decimal places 3→

17Value 4 display→

17

Decimal places 4→18

Display interval→

18

Display damping→

19

Header→20

Header text→

20

Separator→

21

Contrast display→

21

Backlight→

21‣ Configuration backup→

22

Operating time→

22

Last backup→

22

Configuration mgmt.

→

22

Backup state→

23

Comparison result

→

23

→

24

Alarm delay→24‣ Diagnostic
behavior→

25‣ Display‣ Diagnostic handling124

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Diagnostic no. 302→

25

Diagnostic no. 441→

25

Diagnostic no. 444→

25

Diagnostic no.905

→

26

→

26

Device reset→

27

Transmitter identifier→

27

Activate SW option→

28

Software option overview→

28‣ Define access code→

29

Define access code→29

Confirm access code

→

29

→

30

Operating time→

30

Reset access code

→

30

→

31

→

31

→

31

Concentration→

32

Dew point 1→

32

Dew point 2→

32

Cell gas pressure→

32

Cell gas temperature→32

Detect. ref. level→

34

Detect. zero level→

34

Peak 1 index→

34

Peak 1 index delta→

34

Peak 2 index→

34

Peak 2 index delta→

34

Peak track index→

34

Peak track index delta→

34

Midpoint delta

→

34

→

35

‣ Current input 1～n→

35

Measured val. 1～n→35

Measured curr. 1～n→36‣ Val.stat.inp. 1～n→36

Val.stat.inp. 1～n

→

36

→

36

→

36

→

36

‣ Administration‣ Reset access code

‣ Sensor‣ Measured values‣ Measured variables‣ Input values‣ Output values‣ Val. curr.outp 1～n
Output curr.
Endress+Hauser

125機能説明書

J22 TDLAS ガスアナライザ

Measur. curr.

→

36‣ Switch output 1→

37

Switch state

→

37

→

37

Switch state→38

Switch cycles→

38

Max. cycles no.

→

38

→

39

Concentration unit (ppmv)→

39

Temperature unit (°C)→

39

Pressure unit (bar)→40

Length unit (m)→40Date/time format→

40‣ User-specific units→

41

User concentration text→

41

User concentration offset→

41

User concentration factor

→

41

→

42

Analyte type→42

Select calibration→

42

Rolling average number

→

42

→

43

Dew point method 1→

43

Dew point method 2→

43

Conversion type→43

Pipeline pressure mode→

43

Pipeline pressure fixed→

44

Pipeline pressure→

44‣ Calibration 1～n→45Methane CH4→

45

Ethane C2H6→45

Propane C3H8→45

IButane C4H10→45

N-Butane C4H10→45

Isopentane C5H12→45

N-Pentane C5H12→45

Neopentane C5H12→45

Hexane+ C6H14+→

45

Nitrogen N2→

45

Carbon diox. CO2→

45

Hydrog.sulf. H2S→

45‣ Relay output 1～n‣ System units‣ Stream‣ Dew point

126

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Hydrogen H2

→

45

→

46

Peak track analyzer control→

46

Peak track reset→

46

Peak track average number

→

46

→

47

Concentration adjustment→

47

Concentration multiplier→

47

Concentration offset→

47

2f base crv source→

47

2f base RT update→

49

Calibration 1～n→48

Midpoint default→

48

Ramp default→

48

Mod ramp default

→

48

→

49

→

49

Stream change compensation→

51

Methane CH4→51

Ethane C2H6→51

Propane C3H8→

51

IButane C4H10→

51

N-Butane C4H10→51

Isopentane C5H12→

51

N-Pentane C5H12→51

Neopentane C5H12→

51

Hexane+ C6H14+→

51

Nitrogen N2→

51

Carbon diox. CO2→

51

Hydrog.sulf. H2S→

51

Hydrogen H2

→

51

→

51

Det. 1 TIA gain→51

Detector bias→

51

Flow switch input→

51

Flow switch state

→

51

→

52

I/O module 1～n terminals→

52

I/O module 1～n information→

52

I/O module 1～n type→52

Apply I/O configuration→

53‣ Peak tracking‣ Sensor adjustment‣ Stream change compensation (SCC)‣ Calibration 1～n‣ Calibration‣ I/O configuration

Endress+Hauser127

機能説明書

J22 TDLAS ガスアナライザ

I/O alteration code

→

53

→

54

→

54

Terminal no.

→54

Signal mode→

54

Current span→55

0/4 mA value→

55

20 mA value→

55

Failure mode→

56

Failure value

→

56

→

57

→

57

Terminal no.

→57

Signal mode→

63

Proc.var. outp→

63

Curr.range out→

63

Fixed current→

59

Low.range outp→59

Upp.range outp→

60

Damp.curr.outp→

60

Fail.behav.out→

61

Fail. current→

61

Output curr.

→

61

Measur. curr.

→

61

→

63

Signal mode→

63

Operating mode→

63

Switch out funct→

63

Assign diag. beh→

63

Assign limit→

64

Switch-on value→

64

Switch-off value→

65

Assign status→65

Switch-on delay→

65

Switch-off delay→

65

Switch state→

65

Invert outp.sig.

→

65

→

67

Relay output function→

67

Assign limit→

68

Assign diag. beh→

68‣ Input‣ Current input 1～n‣ Output‣ Curr.output 1～n‣ Switch output 1～n‣ Relay output 1～n

128

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Assign status→

68

Switch-off value→

68

Switch-off delay→

68

Switch-on value→

70

Switch-on delay→

70

Switch state→70

Powerless relay

→

70

→

71

‣ Modbus configuration→

71

Bus address→

71

Baudrate→

72

Data transfer mode→

72

Parity→72

Byte order→73

Telegram delay→

74

Prio. IP address→

74

Inactivity timeout→

74

Max connections→

74

Failure mode→75

Bus termination→

75

Fieldbus writing access→

75‣ Modbus information→76

Device ID→

76

Device revision

→

76

→

77

→

77

→

77

Web server language→

77

MAC address→

77

DHCP client→77

IP address→

77

Subnet mask→

79

Default gateway→79

Web server functionality→

79

Login page

→

79

→

81

Actual diagnostics→

81

Previous diagnostics→

82

Operating time from restart→82

Operating time→82

‣ Diagnostic list→

83

‣ Communication‣ Modbus data map
Scan list register 0 to 15‣ Web server‣ Diagnostics

Endress+Hauser

129

機能説明書

J22 TDLAS ガスアナライザ

Diagnostics 1→

83

Diagnostics 2→

83

Diagnostics 3→

83

Diagnostics 4→85

Diagnostics 5

→

85

→

86

→

86

→

87

Device tag→

87

Serial number→

87

Firmware version→

87

Device name→88

Order code→

88

Extended order code 1→

88

Extended order code 2→

88

Extended order code 3→

88

ENP version→

89‣ Main electronic module + I/O module 1→

89

Firmware version→

89

Build no. software→

89

Bootloader revision

→

90

→

90

Firmware version→

90

Build no. software→

90

Bootloader revision

→

90

→

91

I/O module 2 terminal numbers→

91

Firmware version→

91

Build no. software→

91

Bootloader revision

→

91

→

92

I/O module 3 terminal numbers→

92

Firmware version→

92

Build no. software→

92

Bootloader revision

→

92

→

93

Firmware version→

93

Build no. software→

93

Bootloader revision

→

93

→

94

→

94

‣ Event logbook
Filter option‣ Device information‣ Sensor electronic module (ISEM)‣ I/O module 2‣ I/O module 3‣ Display module‣ Data logging
Assign chan. 1～n130

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Logging interval→

94

Clear logging→

95

Data logging→

95

Logging delay→

95

Data log.control→

96

Data log. status→

96

Logging duration

→

97

→

97

→

97

Plant operator→

97

Location→

97‣ Gas validation settings→

98

Select val. cal.

→98

Validation type→

98

Num. val. Points→

99

Val. purge time→99

Meas. duration→

99

Val. gas info→

99

Val. conc.

→

99

Val. allowance→

100

Performing verification

→

100

Year

→

100

Month

→

101

Day

→

101

Hour

→

101

AM/PM

→

102

Minute→102

Meas. Duration→

102

Verification mode→

102

Ext. device info→

103

Start verification→

103

Progress→

103

Status→103

Measured val.

→

104

Output values→

104

Measured conc.

→

104

Verification result

→

104

→

106

Date/time (manually entered)→106

Verification ID→

106

Operating time→106

‣ Heartbeat Technology‣ Heartbeat settings

Verification results

Endress+Hauser

131機能説明書

J22 TDLAS ガスアナライザ

Verification result→

106

Sensor→106

Sens. electronic→106

Gas validation→

106

I/O module→

106

System status

→

106

→

108

Date/time (manually entered)→108

Operating time→108

Gas validation→

108

Concentration average→

108

Conc. std. dev.

→

109

Conc. max→

109

Conc. min

→

109

→

110

Detector reference level→

110

Peak 1 index delta→

110

Peak 2 index delta

→

110

→

110

Curr.inp 1～n sim.

→

110

Value curr.inp1～n→110

Curr.outp1～n sim.

→

111

Curr.outpval. 1～n→111

Switch sim. 1～n→111

Switch state 1～n→

112

Relay out.1～n sim→

112

Switch state 1～n→112

Dev. alarm sim.

→

113

Event category→

113

Diag. event sim.

→

113

→

115

Midpoint default 1～n→114

Ramp default 1～n→115

Concentration→

115

Dew point 1→

115

Dew point 2→

115

Cell gas press.

→

116

Cell gas temp.

→

116

Detect. ref. lvl→

116

Detect. zero lvl→

116

Peak 1 index→

116

Gas validation results

Monitoring results‣ Simulation‣ Spectrum plots

132

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Peak1 idx.delta→

117

Peak 2 index→

117

Peak2 idx.delta→

117

Peak track index→

117

Pk trk idx delta→

118

Midpoint delta→

118

Analyzer control

→

118

Reset

→

118

Det. 1 TIA gain

→

119

→

119

Spectra log rate→

119

Num. spectra file→

119

Val. log level→

120

Num. val. files→

120‣ SD card

6.3 レジスタ情報
ナビゲーション：Expert
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Locking status

4918

整数

読み取り

256 = Hardware locked

10

512 = Temporarily locked
User role

2178

整数

読み取り

0 = Operator

11

1 = MaintenanceEnter access code

2177

整数

読み取り/

4 桁のアクセスコード

11

選択項目/ユーザー入力/ユーザー

→

書き込み

6.3.1

System サブメニュー

6.3.1.1

Display

ナビゲーション：Expert → System → Display
パラメータ

レジスタ

データ型

アクセス

インターフェース
Display language

Format display

Endress+Hauser

3673

3625

整数

整数

読み取り/
書き込み

読み取り/
書き込み

0 = English
1 = Français
2 = Italiano
3 = русский язык (Russian)
4 = 中文 (Chinese)
0 = 1 value, max. size
1 = 1 bargraph + 1 value
2 = 2 values
3 = 1 value large + 2 values
4 = 4 values

13

13

133

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → System → Display
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Value 1 display

3963

整数

0% bargraph value 14136～4137

浮動小数100% bargraph value 14142～4143

浮動小数

Decimal places 1

3365

整数

Value 2 display

Decimal places 2

3964

4049

整数

整数

Value 3 display

3966

整数

0% bargraph value 34138～4139

浮動小数100% bargraph value 3

4140～4141

浮動小数

Decimal places 3

4050

整数

読み取り/
書き込み

読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み

読み取り/
書き込み
読み取り/
書き込み

読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み

2 = Cell gas pressure
3 = Cell gas temperature
4 = Dew point 1
5 = Dew point 2
151 = Concentration
符号付き浮動小数点数

14

14

符号付き浮動小数点数

140=x
1 = x.x
2 = x.xx
3 = x.xxx
4 = x.xxxx
選択リストについては、Value 1
display パラメータ（→ 14）を参照
0=x
1 = x.x
2 = x.xx
3 = x.xxx
4 = x.xxxx
選択リストについては、Value 1
display を参照
符号付き浮動小数点数
符号付き浮動小数点数

15

16

16

17
17
17

0=x
1 = x.x
2 = x.xx

17

3 = x.xxx
4 = x.xxxx
Value 4 display

3965

整数

読み取り/
書き込み

選択リストについては、Value 1
display を参照

Decimal places 4

4051

整数

読み取り/
書き込み

0=x

17

1 = x.x
2 = x.xx

18

3 = x.xxx
4 = x.xxxx
Display interval

3604～3605

浮動小数

Display damping

3554～3555

浮動小数

134

読み取り/
書き込み
読み取り/
書き込み

1～10 s
0.0～999.9 s

18
19

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ナビゲーション：Expert → System → Display
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Header3624

整数

Header text

3968～3973

文字列Separator

3671

整数

Contrast display

3674～3675

浮動小数

Backlight

3967

整数

6.3.1.2

読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み

0 = Disable
1 = Enable

20
20
21
21
21

Configuration backup

ナビゲーション：Expert → System → Configuration backup
パラメータ
レジスタ
データ型
アクセス
Operating time
Last backup
Configuration
management

2631～2637
6430～6436
5500

文字列
文字列
整数

読み取り
読み取り
読み取り/
書き込み

Backup state

5502

整数

読み取り

Comparison result

5514

整数

読み取り

6.3.1.3

0 = Device tag
1 = Free text
最大 12 文字（英字、数字、特殊文字（例：@、%、/）など）
1 = .（点）2 = ,（コンマ）20～80 %

選択項目/ユーザー入力/ユーザー
インターフェース
日（d）、時（h）、分（m）、秒（s）
日（d）、時（h）、分（m）、秒（s）
0 = Cancel
1 = Execute backup
2 = Restore
4 = Clear backup data
5 = Compare
1 = Backup in progress
2 = Restoring in progress
4 = Delete in progress
5 = Compare in progress
6 = Restoring failed
7 = Backup failed
251 = None
0 = Settings identical
1 = Settings not identical
2 = No backup available
3 = Check not done
4 = Backup settings corrupt
5 = Dataset incompatible→

選択項目/ユーザー入力/ユーザー
インターフェース
0～60 s

→

22
22
22

23

23

Diagnostic handling

ナビゲーション：Expert → System → Diagnostic handling
パラメータ
レジスタ
データ型
Alarm delay

Endress+Hauser

6808～6809

浮動小数

アクセス
読み取り/
書き込み

24

135

機能説明書

J22 TDLAS ガスアナライザ

Diagnostic behavior サブメニュー
ナビゲーション：Expert → System → Diagnostic handling → Diagnostic behavior
パラメータ
レジスタ
データ型
アクセス 選択項目/ユーザー入力/ユーザー
インターフェース
Diagnostic no. 302
2312
整数
読み取り/ 0 = Warning
書き込み
1 = Alarm
Diagnostic no. 441
4742
整数
読み取り/ 0 = Off
書き込み
1 = Logbook entry only
2 = Warning
3 = Alarm
Diagnostic no. 444
5120
整数
読み取り/ 0 = Off
書き込み
1 = Logbook entry only
2 = Warning
3 = Alarm
Diagnostic no. 905
30025
整数
読み取り/ 0 = Off
書き込み
1 = Alarm
2 = Warning
3 = Logbook entry only
4 = Reset

6.3.1.4

→
25

25

25

26

Administration

ナビゲーション：Expert → System → Administration
パラメータ
レジスタ
データ型

アクセス

Device reset

6817

整数

読み取り/
書き込み

Transmitter identifier
Activate SW option

4510
2795

整数
文字列

Software option
overview

2902

整数

読み取り
読み取り/
書き込み
読み取り

選択項目/ユーザー入力/ユーザー
インターフェース
0 = Cancel
1 = Restart device
2 = To delivery settings
1 = 300
最大 10 桁の数字→

1 = Extended HistoROM
32768 = Heartbeat Verification
16384 = Heartbeat Monitoring

28

27

27
28

Define access code サブメニュー
ナビゲーション：エキスパート → システム → 管理 → アクセスコード設定
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Define access code
Confirm access code

136

8677～8684
8685～8692

文字列
文字列

読み取り/
書き込み

数字、英字、特殊文字から成る最

読み取り/
書き込み

数字、英字、特殊文字から成る最29

大 16 桁の文字列29

大 16 桁の文字列

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Reset access code サブメニュー
ナビゲーション：Expert → System → Administration → Reset access code
パラメータ
レジスタ
データ型
アクセス 選択項目/ユーザー入力/ユーザー
インターフェース
Operating time
2631～2637
文字列
読み取り 日（d）、時（h）、分（m）、秒（s）
Reset access code8880～8895
文字列
読み取り/ 数字、英字、特殊文字から成る文
書き込み
字列

6.3.2

Sensor6.3.2.1

Measured values

→
30
30

Measured variables サブメニュー
ナビゲーション：Expert → Sensor → Measured values → Measured variables
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Concentration

9455～9456

浮動小数

読み取り

0～1000000 ppmv

32

Dew point 1

21458～21459

浮動小数

読み取り

符号付き浮動小数点数

32

Dew point 2

21800～21801

浮動小数

読み取り

符号付き浮動小数点数

32

Cell gas pressure25216～25217

浮動小数

読み取り

符号付き浮動小数点数

33

Cell gas temperature

21854～21855

浮動小数

読み取り

符号付き浮動小数点数

33

Detector reference level4720～4721

浮動小数

読み取り

0～5 mA

33

Detector zero level9667～9668

浮動小数

読み取り

0～5 mA

34

Peak 1 index

9834～9835

浮動小数

読み取り

0.0～511.0

34

Peak 1 index delta

30581～30582

浮動小数

読み取り

-511.0～+511.0

34

Peak 2 index

27600～27601

浮動小数

読み取り

0.0～511.0

34

Peak 2 index delta

30672～30673

浮動小数

読み取り

-511.0～+511.0

34

Peak track index

29018～29019

浮動小数

読み取り

0.0～511.0

34

Peak track index delta

28814～28815

浮動小数

読み取り

-511.0～+511.0

34

Midpoint delta

47236～47237

浮動小数

読み取り

0.0～120.0 mA

34

Input values サブメニュー

Current input 1～n
ナビゲーション：Expert → Sensor → Measured values → Input values → Current input 1～n
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Measured values 1～n

1: 6151～6152

浮動小数

読み取り

符号付き浮動小数点数

2: 6153～6154

35

3: 6155～6156
Measured current 1～n

1: 6131～6132
2: 6133～6134

浮動小数

読み取り

0～22.5 mA36

3: 6135～6136

Value status input 1～nEndress+Hauser137

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → Sensor → Measured values → Input values → Value status input 1～n
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Value status input 1～n

1: 2746

整数

読み取り2: 4699

0 = Low
1 = High36

3: 4700

Output values サブメニュー

Value current output 1～n
ナビゲーション：Expert → Sensor → Measured values → Output values → Value current output 1～n
パラメータ
Output current 1～n

Measured current 1～n

レジスタ
1: 5931～5932
2: 5933～5934
3: 5935～5936
1: 5779～5780
2: 5781～5782
3: 5783～5784

データ型
浮動小数

アクセス

選択項目/ユーザー入力/ユーザー

読み取り

インターフェース
0～22.5 mA

→

36
浮動小数

読み取り

0～30 mA
36

Switch output 1～n
ナビゲーション：Expert → Sensor → Measured values → Output values → Switch output 1～n
パラメータ
Switch state 1～n

レジスタ
1: 2485
2: 2486
3: 9917

データ型
整数

アクセス

選択項目/ユーザー入力/ユーザー

→

読み取り

インターフェース
1 = Open
6 = Closed

37

Relay output 1～n
ナビゲーション：Expert → Sensor → Measured values → Output values → Relay output 1～n
パラメータ
Switch state

Switch cycles

Max. switch cycles
number6.3.2.2

レジスタ
1: 3518
2: 3519
3: 9875
1: 7625
2: 7627
3: 7629
1: 21919
2: 21921
3: 21923

データ型
整数

整数

選択項目/ユーザー入力/ユーザー

→

読み取り

インターフェース
1 = Open
6 = Closed

38

読み取り

正の整数
38

整数

読み取り

正の整数
38

System units

ナビゲーション：Expert → Sensor → System units
パラメータ
レジスタ
データ型
Concentration unit

138

アクセス

2439

整数

アクセス
読み取り/
書き込み

選択項目/ユーザー入力/ユーザー
インターフェース
0 = ppmv
1 = lb/MMscf→
39

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ナビゲーション：Expert → Sensor → System units
パラメータ
レジスタ
データ型

Temperature unit

Pressure unit

Length unit

Date/time format

2109

2130

2087

2150

整数

整数

整数

整数

アクセス

読み取り/
書き込み

読み取り/
書き込み

読み取り/
書き込み

読み取り/
書き込み

選択項目/ユーザー入力/ユーザー
インターフェース
2 = %vol3 = mg/sm3
4 = ppbv
5 = mg/Nm3
240 = User conc.
0 = °C
1=K
2 = °F
3 = °R
0 = bar
1 = psi a
2 = bar g
3 = psi g
4 = Pa a
5 = kPa a
6 = MPa a
7 = Pa g
8 = kPa g
9 = MPa g
44 = ft
45 = m47 = in
49 = mm
240 = μm
0 = dd.mm.yy hh:mm
1 = mm/dd/yy hh:mm am/pm
2 = dd.mm.yy hh:mm am/pm
3 = mm/dd/yy hh:mm

→

39

40

40

40

User-specific units サブメニュー
ナビゲーション：Expert → Sensor → System units → User-specific units
パラメータ
レジスタ
データ型
アクセス 選択項目/ユーザー入力/ユーザー
インターフェース
User concentration text2585～2589
文字列
読み取り/ 最大 10 文字（英字、数字、特殊書き込み
文字（@、%、/）など）
User concentration
2490～2491
浮動小数読み取り/ 符号付き浮動小数点数
offset
書き込み
User concentration
2554～2555
浮動小数読み取り/ 符号付き浮動小数点数
factor書き込み

6.3.2.3

→
41
41
41

Streamナビゲーション：Expert → Sensor → Stream
パラメータ
レジスタ
データ型
Analyte type

Endress+Hauser

21930

整数

アクセス
読み取り/
書き込み

選択項目/ユーザー入力/ユーザー
インターフェース
0 = H2O
1 = CO2
2 = H2S
3 = CH4
4 = NH3

→

42

139

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → Sensor → Stream
パラメータ
レジスタ
データ型

Select calibration

Rolling average number

6.3.2.4

22968

6876

整数

整数

読み取り/
書き込み

読み取り/
書き込み

選択項目/ユーザー入力/ユーザー
インターフェース
5 = HCl
6 = O2
7 = CO
8 = SO2
9 = C2H2
0=1
1=2
2=3
3=4
1～256

→

選択項目/ユーザー入力/ユーザー
インターフェース
0 = Off
1 = ASTM1
2 = ASTM2
3 = ISO
4 = AB
0 = Off
1 = ASTM1
2 = ASTM2
3 = ISO
4 = AB
0 = Ideal
1 = Real
1 = Fixed value
0 = External value
11 = Current input 1
12 = Current input 2
13 = Current input 3
符号付き浮動小数点数

→

42

42

Dew Point

ナビゲーション：Expert → Sensor → Dew point
パラメータ
レジスタ
データ型
Dew point method 1

Dew point method 2

21595

7631

整数

整数

Conversion type

21596

整数

Pipeline pressure mode

48175

整数

Pipeline pressure fixed48251～48252

浮動小数

Pipeline pressure9483～9484

浮動小数

140

アクセス

アクセス
読み取り/
書き込み

読み取り/
書き込み

読み取り/
書き込み
読み取り/
書き込み

読み取り/
書き込み
読み取り/
書き込み

符号付き浮動小数点数

43

43

43

43

44
45

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Dew point calibration サブメニュー
ナビゲーション：Expert → Sensor → Dew Point → Calibration 1～n
パラメータ
Methane CH4
Ethane C2H6
Propane C3H8
Isobutane C4H10
N-Butane C4H10
Isopentane C5H12
N-Pentane C5H12Neopentane C5H12
Hexane+ C6H14+
Nitrogen N2
Carbon diox. CO2
Hydrog.sulf. H2S
Hydrogen H2

6.3.2.5

レジスタ

データ型

アクセス

26445, 26453,
26461, 26469
26317, 26325,
26333, 26341
26509, 26517,
26525, 26533
25486, 25494,
25502, 25510
26915, 26917,
26919, 26921
27968, 27970,
27972, 27974
26931, 26933,
26935, 26937
26923, 26925,
26927, 26929
27976, 27978,
27980, 27982
25314, 25322,
25330, 25338
26199, 26207,
26215, 26223
26381, 26389,
26397, 26405
29191, 29193,
29195, 29197

浮動小数

読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み

浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数
浮動小数

選択項目/ユーザー入力/ユーザー
インターフェース
0.4～1.0 モル分率
0.0～0.2 モル分率
0.0～0.15 モル分率
0.0～0.1 モル分率
0.0～0.1 モル分率
0.0～0.1 モル分率
0.0～0.1 モル分率
0.0～0.1 モル分率
0.0～0.1 モル分率
0.0～0.55 モル分率
0.0～0.3 モル分率
0.0～0.05 モル分率
0.0～0.2 モル分率

→
45
45
45
45
45
45
45
45
45
45
45
45
45

Peak Tracking

ナビゲーション：Expert → Sensor → Peak tracking
パラメータ

レジスタ

データ型

アクセス

Peak track analyzer
controlPeak track reset

21460

整数

4727

整数

Peak track average
number

21568

整数

読み取り/
書き込み
読み取り/
書き込み
読み取り/
書き込み

6.3.2.6

選択項目/ユーザー入力/ユーザー
インターフェース
0 = Off
1 = On0 = Off
3 = Reset
1～3600→

選択項目/ユーザー入力/ユーザー
インターフェース
0 = Off
1 = On-1000000～+1000000

→

46
46
46

Sensor adjustment

ナビゲーション：Expert → Sensor → Sensor adjustment
パラメータ

レジスタ

データ型

アクセス

Concentration
adjustment
Concentration
multiplier

47129

整数

47222～47223

浮動小数

読み取り/
書き込み
読み取り/
書き込み

Endress+Hauser

47
47

141

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → Sensor → Sensor adjustment
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Concentration offset

47224～47225

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

2f base crv source

28614

整数

読み取り/
書き込み

0 = Ref0 curve
1 = Ref0 RT curve47

2f base RT update

30669

整数

読み取り/
書き込み

0 = Cancel
1 = Start

49

48

Sensor adjustment calibration サブメニュー
ナビゲーション：Expert → Sensor → Sensor adjustment → Calibration
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Laser midpoint default

31090, 31092,
31094, 31096

浮動小数

読み取り/
書き込み

0～120 mA

Laser ramp default

26750, 26752,
26754, 26756

浮動小数

読み取り/
書き込み

0～120 mA

Laser modulation
amplitude default

36077, 36079,
36081, 36083

浮動小数

読み取り/
書き込み

0～100 mA

6.3.2.7

→
48
48
48

Stream change compensation (SCC)

ナビゲーション：Expert → Sensor → Stream change compensation (SCC)
パラメータ

レジスタ

データ型

アクセス

Calibration 1～n35689～35692

整数

読み取り

選択項目/ユーザー入力/ユーザー
インターフェース
1 = No0 = Yes→49

SCC calibration サブメニュー
ナビゲーション：Expert → Sensor → Sensor adjustment → Calibration (1～n)
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Methane CH426445, 26453,
26461, 26469

浮動小数

読み取り/
書き込み

0.4～1.0 モル分率

Ethane C2H626317, 26325,
26333, 26341

浮動小数

読み取り/
書き込み

0.0～0.2 モル分率

Propane C3H826509, 26517,
26525, 26533

浮動小数

読み取り/
書き込み

0.0～0.15 モル分率

Isobutane C4H1025486, 25494,
25502, 25510

浮動小数

読み取り/
書き込み

0.0～0.1 モル分率

N-Butane C4H10

26915, 26917,
26919, 26921

浮動小数

読み取り/
書き込み

0.0～0.1 モル分率

Isopentane C5H1227968, 27970,
27972, 27974

浮動小数

読み取り/
書き込み

0.0～0.1 モル分率

N-Pentane C5H1226931, 26933,
26935, 26937

浮動小数

読み取り/
書き込み

0.0～0.1 モル分率

142

→
51
51
51
51
51
51
51

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ナビゲーション：Expert → Sensor → Sensor adjustment → Calibration (1～n)
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Neopentane C5H1226923, 26925,
26927, 26929

浮動小数

読み取り/
書き込み

0.0～0.1 モル分率

Hexane+ C6H14+

27976, 27978,
27980, 27982

浮動小数

読み取り/
書き込み

0.0～0.1 モル分率

Nitrogen N225314, 25322,
25330, 25338

浮動小数

読み取り/
書き込み

0.0～0.55 モル分率

Carbon diox. CO2

26199, 26207,
26215, 26223

浮動小数

読み取り/
書き込み

0.0～0.3 モル分率

Hydrog.sulf. H2S

26381, 26389,
26397, 26405

浮動小数

読み取り/
書き込み

0.0～0.05 モル分率

Hydrogen H229191, 29193,
29195, 29197

浮動小数

読み取り/
書き込み

0.0～0.2 モル分率

6.3.2.8

→
51
51
51
51
51
51

Calibration

ナビゲーション：Expert → Sensor → Calibration
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザーインターフェース→

Det. 1 TIA gain

29235

整数

読み取り

0～15

51

Detector bias

29237～
29238

浮動小数

読み取り/
書き込み

符号付き浮動小数点値

Flow switch input

4712

整数

読み取り/
書き込み

0 = Off
1 = Normally open

51

51

2 = Normally closed
Flow switch state

29222

整数

読み取り

0 = No flow
1 = Flow

6.3.3

51

I/O configuration サブメニュー

ナビゲーション：Expert → I/O configuration
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

I/O module 1～nterminal numbers

1: 6541

整数

読み取り

0 = Not used

2: 6542

1 = 26-27 (I/O 1)

3: 6543

2 = 24-25 (I/O 2)→

52

3 = 22-23 (I/O 3)
I/O module 1～ninformation

1: 8659

整数

読み取り

1 = MODBUS

2: 8660

2 = Configurable

3: 8661

3 = Not configurable

52

254 = Not plugged
255 = Invalid

Endress+Hauser143

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → I/O configuration
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

I/O module 1～n type

1: 6417

整数

読み取り/
書き込み

0 = Off

2: 6418
3: 6419

1 = Current output
2 = Current input

I/O alteration code

8665
6427

6.3.4

Input サブメニュー

6.3.4.1

Current input 1～n

整数
整数

読み取り/
書き込み

0 = Yes

読み取り/
書き込み

正の整数

52

1

1

6 = Relay output
Apply I/O configuration

1

1

3 = Switch output
5 = Status input

→

1

52

1 = No53

ナビゲーション：Expert → Input → Status input 1～n
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Terminal number

1: 6548

整数

読み取り

0 = Not used

2: 6549

1 = 26-27 (I/O 1)

3: 6550

2 = 24-25 (I/O 2)→

54

3 = 22-23 (I/O 3)
Signal mode

1: 6424

整数

2: 6425
Current span

1: 6147

整数

2: 6148

読み取り/
書き込み

0 = Passive

読み取り/
書き込み

0 = 4 to 20 mA (4 to 20.5 mA)54

2 = Active
1 = 4 to 20 mA US (3.9 to 20.8 mA)
2 = 4 to 20 mA NAMUR (3.8 to 20.5mA)

55

3 = 0 to 20 mA (0 to 20.5 mA)
0/4 mA value

1: 6111～6112

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

整数

読み取り/
書き込み

1 = Last valid value

2: 6113～6114
20 mA value

1: 6119～6120
2: 6121～6122

Failure mode

1: 6159
2: 6160

2 = Alarm

55
55

56

6 = Defined value
Failure value

1: 6163～6164
2: 6165～6166

1

浮動小数

読み取り/
書き込み

符号付き浮動小数点数56

表示項目は注文オプションまたは機器設定に応じて異なります。

144

Endress+Hauser

J22 TDLAS ガスアナライザ

6.3.5

Output サブメニュー

6.3.5.1

Current output 1～n

機能説明書

ナビゲーション：Expert → Output → Current output 1～n
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザーイ
ンターフェース

Terminal number

1: 6545

整数

読み取り

0 = Not used

2: 6546

1 = 26-27 (I/O 1)
2 = 24-25 (I/O 2)→

57

3 = 22-23 (I/O 3)
Signal mode

1: 6421整数

2: 6422
Process variablecurrent output

5927～5929

整数

読み取り/
書き込み

0 = Passive

読み取り/
書き込み

0 = Off

2 = Active

63

151 = Concentration
3 = Cell gas temperature

63

4 = Dew point 1
5 = Dew point 2
Current range output

1: 5923

整数

2: 5924

読み取り/
書き込み

0 = 4 to 20 mA (4 to 20.5 mA)
1 = 4 to 20 mA US (3.9 to 20.8 mA)
2 = 4 to 20 mA NAMUR (3.8 to 20.5mA)

63

3 = 0 to 20 mA (0 to 20.5 mA)
4 = Fixed value
Fixed current

1: 5987～5988

浮動小数

読み取り/
書き込み

0～22.5 mA

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

浮動小数

読み取り/
書き込み

0.0～999.9 s

整数

読み取り/
書き込み

0 = Min.

2: 5989～5990
Lower range value
output

1: 6195～6196

Upper range value
output

1: 5915～5916

Damping current
output

1: 5903～5904

Failure behavior
current output

1: 5911

2: 6197～6198
2: 5917～5918
2: 5905～5906
2: 5912

59
59
60
60

1 = Max.
4 = Actual value

61

5 = Last valid value
6 = Fixed value
Failure current

1: 5979～5980

浮動小数

読み取り/
書き込み

0～22.5 mA

浮動小数

読み取り

0～22.5 mA

浮動小数

読み取り

0～30 mA

2: 5981～5982
Output current 1～n

1: 5931～5932
2: 5933～5934

Measured current 1～n

1: 5779～5780
2: 5781～5782

Endress+Hauser

61
61
61

145

機能説明書6.3.5.2

J22 TDLAS ガスアナライザ

Switch output 1～n

ナビゲーション：Expert → Output → Switch output 1～n
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Terminal number

1: 6551

整数

読み取り

0 = Not used

2: 6552

1 = 26-27 (I/O 1)
2 = 24-25 (I/O 2)→

63

3 = 22-23 (I/O 3)
Signal mode

1: 6235

整数

2: 6236

読み取り/
書き込み

0 = Passive
2 = Active

63

3 = Passive NAMUR
Operating mode

1: 4479

整数

読み取り/
書き込み

1 = Switch

整数

読み取り/
書き込み

0 = Off

2: 4480
Switch output function

1: 3022
2: 302363

1 = On2 = Diagnostic behavior

63

4 = Limit
5 = Status
Assign diagnostic
behavior

1: 3096

整数

2: 3097

読み取り/
書き込み

0 = Alarm
1 = Warning64

2 = Alarm or warning
Assign limit

1: 3184

整数

2: 3185

読み取り/
書き込み

0 = Off
151 = Concentration
4 = Dew point 1

64

5 = Dew point 2
Switch-on value

1: 3242～3243

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

浮動小数

読み取り/
書き込み

0.0～100.0 s

浮動小数

読み取り/
書き込み

0.0～100.0 s

整数

読み取り/
書き込み

0 = Actual status

2: 3244～3245
Switch-off value

1: 3234～3235
2: 3236～3237

Switch-on delay

1: 6247～6248
2: 6249～6250

Switch-off delay

1: 6239～6240
2: 6241～6242

Failure mode

1: 3384
2: 3385

1 = Open

65
65
65
65

65

6 = Closed
Switch state 1～n

1: 2485

整数

読み取り

2: 2486
Invert output signal

1: 2583
2: 2584

146

1 = Open
6 = Closed

整数

読み取り/
書き込み

0 = Yes
1 = No65
65

Endress+Hauser

J22 TDLAS ガスアナライザ

6.3.5.3

機能説明書

Relay output 1

ナビゲーション：Expert → Output → Relay output 1～n
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Terminal number

1: 8278
2: 8279

整数

読み取り

0 = Not used
1 = 26-27 (I/O 1)
2 = 24-25 (I/O 2)
3 = 22-23 (I/O 3)67

1: 2488
2: 2489

整数

1 = Open
2 = Diagnostic behavior
4 = Limit
5 = Status
6 = Closed

67

読み取り/
書き込み

0 = Off
4 = Dew point 1
5 = Dew point 2
151 = Concentration

68

読み取り/
書き込み

0 = Alarm
1 = Warning
2 = Alarm or warning

68

Relay output function

Assign limit

読み取り/
書き込み

→

1: 8248
2: 8249

整数

Assign diagnostic
behavior

1: 8245
2: 8246

整数

Switch-off value

1: 8260～8261
2: 8262～8263

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

Switch-off delay

1: 8254～8255
2: 8256～8257

浮動小数

読み取り/
書き込み

0.0～100.0 s

Switch-on value

1: 8233～8234
2: 8235～8236

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

Switch-on delay

1: 8266～8267
2: 8268～8269

浮動小数

読み取り/
書き込み

0.0～100.0 s

Failure mode

1: 8242
2: 8243

整数

読み取り/
書き込み

0 = Actual status
1 = Open
6 = Closed

70

Switch state

1: 3518
2: 3519

整数

読み取り

1 = Open
6 = Closed

70

Powerless relay status

1: 7009
2: 7010

整数

読み取り/
書き込み

1 = Open
6 = Closed

70

6.3.6

Communication サブメニュー

6.3.6.1

Modbus configuration

68
68
68
70

ナビゲーション：Expert → Communication → Modbus configuration
パラメータ
Bus address

1

1

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

4910

整数

読み取り/
書き込み

1～247

→
71

Modbus RS485 のみ

Endress+Hauser

147

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → Communication → Modbus configuration
パラメータ
1

Baudrate

Data transfer mode
Parity1

1

Byte order

Telegram delay

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

4912

整数

読み取り/
書き込み

0 = 1200 BAUD
1 = 2400 BAUD
2 = 4800 BAUD
3 = 9600 BAUD
4 = 19200 BAUD
5 = 38400 BAUD
6 = 57600 BAUD
7 = 115200 BAUD

72

1

Inactivity timeout
Max connections

2

2

2

Failure mode
Bus termination

1

Fieldbus writing access

→

4913

整数

読み取り/
書き込み

0 = RTU
1 = ASCII

72

4914

整数

読み取り/
書き込み

0 = Even
1 = Odd
2 = None / 2 stop bits
3 = None / 1 stop bit

72

読み取り/
書き込み

0 = 0-1-2-3
1 = 3-2-1-0
2 = 2-3-0-1
3 = 1-0-3-2

73

4915

Priority IP address

6.3.6.2

レジスタ

整数

4916～4917

浮動小数

読み取り/
書き込み

0～100 ms

28273～28280

文字列

読み取り/
書き込み

4 オクテット：0～255（特定のオクテットにおいて）47014～47015

浮動小数

読み取り/
書き込み

0～99 s

47016

整数

読み取り/
書き込み

1～4

4920

整数

読み取り/
書き込み

1 = Last valid value
3
255 = NaN value

75

5774

整数

読み取り

0 = Off
1 = On

75

6807

整数

読み取り/
書き込み

0 = Read + write
1 = Read only

75

74
74
74
74

Modbus information

ナビゲーション：Expert → Communication → Modbus information
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Device ID

2547

整数

読み取り

4 桁の 16 進数

76

Device revision

4481

整数

読み取り

4 桁の 16 進数

761

Modbus RS485 のみ

2

Modbus TCP のみ

3

NaN = 非数

148

Endress+Hauser

J22 TDLAS ガスアナライザ

6.3.6.3

機能説明書

Modbus data map

ナビゲーション：Expert → Communication → Modbus data map
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Scan list register 0 to 15

0: 5001
1: 5002
2: 5003
3: 5004
4: 5005
5: 5006
6: 5007
7: 5008
8: 5009
9: 5010
10: 5011
11: 5012
12: 5013
13: 5014
14: 5015
15: 5016

整数

読み取り/
書き込み

1～65,535

77

Scan list data area 0 to
15

0: 5051～5052
1: 5053～5054
2: 5055～5056
3: 5057～5058
4: 5059～5060
5: 5061～5062
6: 5063～5064
7: 5065～5066
8: 5067～5068
9: 5069～5070
10: 5071～5072
11: 5073～5074
12: 5075～5076
13: 5077～5078
14: 5079～5080
15: 5081～5082

整数/
浮動小数

読み取り/
書き込み

入力したスキャンリストレジスタに応じて異なる

77→

6.3.6.4

Web server

ナビゲーション：Expert → Communication → Web server
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Web server language

4219

整数

読み取り/
書き込み

0 = English
1 = Français
2 = Italiano
3 = русский язык (Russian)
4 = 中文 (Chinese)77

MAC address

4210～4218

文字列

読み取り

英字と数字から成る一意的な 12
桁の文字列

77

DHCP client

21781

整数

読み取り/
書き込み

0 = Off
1 = On77

Endress+Hauser

149

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → Communication → Web server
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

IP address

4155～4162

文字列

読み取り/
書き込み

4 オクテット：0～255（特定のオクテットにおいて）

77

Subnet mask4163～4170

文字列

読み取り/
書き込み

4 オクテット：0～255（特定のオクテットにおいて）

79

Default gateway

4171～4178

文字列

読み取り/
書き込み

4 オクテット：0～255（特定のオクテットにおいて）

79

Web server
functionality

4220

整数

読み取り/
書き込み

0 = Off
1 = On2 = HTML Off

79

Login page

5802

整数

読み取り/
書き込み

0 = Without header
1 = With header

79

6.3.7

→

Diagnostics

ナビゲーション：Expert → Diagnostics
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Actual diagnostic status
signal

2075

整数

読み取り

0: OK
1: Failure (F)2: Function check (C)8: Out of specification (S)4: Maintenance required (M)
16: --- 32: 未分類→

81

Actual diagnostic
number

6801

整数

読み取り

0～65,535

Actual diagnostic
service ID

2732

整数

読み取り

0～65,535

Actual diagnostic string

6821～6830

文字列

読み取り

診断番号、サービス ID、ステー
タス信号

Previous diagnostics
service ID

2734

整数

読み取り

0～65,535

Operating time fromrestart2624～2630

文字列

読み取り

日（d）、時（h）、分（m）、秒（s）

Operating time

2631～2637

文字列

読み取り

日（d）、時（h）、分（m）、秒（s）

82

6.3.7.1

81
81
81
82
82

Diagnostic list

ナビゲーション：Expert → Diagnostics → Diagnostic list
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Diagnostics 1

2736

整数

読み取り

0～65,535

83

Diagnostics 2

2738

整数

読み取り

0～65,535

83

Diagnostics 3

2740

整数

読み取り

0～65,535

83

Diagnostics 4

2742

整数

読み取り

0～65,535

85

Diagnostics 5

2744

整数

読み取り

0～65,53585

150

Endress+Hauser

J22 TDLAS ガスアナライザ

6.3.7.2

機能説明書

Event logbook

ナビゲーション：Expert → Diagnostics → Event logbook
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Filter options

4596

整数

読み取り/
書き込み

0 = Failure (F)86

4 = Maintenance required (M)8 = Function check (C)12 = Out of specification (S)16 = Information (I)255 = All6.3.7.3

Device information

ナビゲーション：Expert → Diagnostics → Device information
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Device tag

2026～2041

文字列

読み取り

最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

87

Serial number

7003～7008

文字列

読み取り

最大 11 文字の英字および数字87

Firmware version

7277～7280

文字列

読み取り

形式 xx.yy.zz の文字列

87

Device name7238～7245

文字列

読み取り

J22 TDLAS ガスアナライザ

88

Order code

2058～2067

文字列

読み取り

英字、数字、特定の句読記号（/など）で構成される文字列

88

Extended order code 1

2212～2221

文字列

読み取り

文字列

88

Extended order code 22222～2231

文字列

読み取り

文字列

88

Extended order code 32232～2241

文字列

読み取り

文字列

88

ENP version

4003～4010

文字列

読み取り

文字列

89

6.3.7.4→

Main electronic module + I/O module 1

ナビゲーション：Expert → Diagnostics → Main electronic module + I/O module 1
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Firmware version

7039

整数

読み取り

正の整数

89

Build no. software

2326

整数

読み取り

正の整数

89

Bootloader revision

2264

整数

読み取り

正の整数

90

6.3.7.5

Sensor electronic module (ISEM)

ナビゲーション：Expert → Diagnostics → Sensor electronic module (ISEM)
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Firmware version

5165

整数

読み取り

正の整数

90

Build no. software

4989

整数

読み取り

正の整数

90

Bootloader revision

4802

整数

読み取り

正の整数

90

Endress+Hauser151

機能説明書6.3.7.6

J22 TDLAS ガスアナライザ

I/O module 2

ナビゲーション：Expert → Diagnostics → I/O module 2
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユー
ザーインターフェース

I/O module 2 terminal
numbers

6542

整数

読み取り

0 = Not used
1 = 26-27 (I/O 1)→

91

2 = 24-25 (I/O 2)
3 = 22-23 (I/O 3)
Firmware version

9877

整数

読み取り

正の整数

91

Build no. software

9918

整数

読み取り

正の整数

91

Bootloader revision

9984

整数

読み取り

正の整数

91

6.3.7.7

I/O module 3

ナビゲーション：Expert → Diagnostics → I/O module 3
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

I/O module 3 terminal
numbers

6543

整数

読み取り

0 = Not used
1 = 26-27 (I/O 1)→

92

2 = 24-25 (I/O 2)
3 = 22-23 (I/O 3)
Firmware version

9879

整数

読み取り

正の整数

92

Build no. software

9919

整数

読み取り

正の整数

92

Bootloader revision

9986

整数

読み取り

正の整数

92

6.3.7.8

Display module

ナビゲーション：Expert → Diagnostics → Display module
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Firmware version

5163

整数

読み取り

正の整数

93

Build no. software

4988

整数

読み取り

正の整数

93

Bootloader revision

4800

整数

読み取り

正の整数

93

6.3.7.9

Data logging

ナビゲーション：Expert → Diagnostics → Data logging
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユー
ザーインターフェース

Assign chan. 1

2445

整数

読み取り/
書き込み

0 = Off→

2 = Cell gas pressure
3 = Cell gas temperature
4 = Dew point 1
5 = Dew point 2

94

121 = Current output 1
122 = Current output 2
151 = Concentration
152 = Flow switch state
152

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ナビゲーション：Expert → Diagnostics → Data logging
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユー
ザーインターフェース

Assign chan. 2

2446

整数

読み取り/
書き込み

0 = Off
2 = Cell gas pressure
3 = Cell gas temperature
4 = Dew point 1
5 = Dew point 2
121 = Current output 1
122 = Current output 2
151 = Concentration
152 = Flow switch state94

0 = Off
2 = Cell gas pressure
3 = Cell gas temperature
4 = Dew point 1
5 = Dew point 2
121 = Current output 1
122 = Current output 2
151 = Concentration
152 = Flow switch state94

0 = Off
2 = Cell gas pressure
3 = Cell gas temperature
4 = Dew point 1
5 = Dew point 2
121 = Current output 1
122 = Current output 2
151 = Concentration
152 = Flow switch state

94

Assign chan. 3

Assign chan. 4

2548

4286

整数

整数

読み取り/
書き込み

読み取り/
書き込み

→

Logging interval

4288～
4289

浮動小数

読み取り/
書き込み

0.1～3600.0 s

Clear logging

4287

整数

読み取り/
書き込み

0 = Cancel
2 = Clear data

95

Data logging

5950

整数

読み取り/
書き込み

0 = Overwriting
1 = Not overwriting

95

Logging delay

5938

整数

読み取り/
書き込み

0～999 時間

Data logging control

5930

整数

読み取り/
書き込み

0 = None
1 = Stop
2 = Delete + start

96

読み取り/
書き込み

0 = Done
1 = Stopped
2 = Active
3 = Delay active

96

読み取り/
書き込み

浮動小数点を含む正の数

Data logging status

Logging duration

Endress+Hauser

5937

2827～
2828

整数

浮動小数

94

95

97

153

機能説明書

J22 TDLAS ガスアナライザ

6.3.7.10 Heartbeat Technology

Heartbeat settings サブメニュー
ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Heartbeat settings
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Plant operator

3414～3429

文字列

読み取り/
書き込み

最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

97

Location

3430～3445

文字列

読み取り/
書き込み

最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

97

Gas validation settings サブメニュー
ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Heartbeat settings → Gas validation settings
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Select validation
calibration

4717

整数

読み取り/
書き込み

0=1
1=2

→

98

2=3
3=4
Validation type

1

26456

整数
整数

読み取り/
書き込み

0 = Validation manual gas

読み取り/
書き込み

0=1

98

1 = Validation auto gas

Number of validation
points

30005

Validation gas purge
time33276～
33277

浮動小数

読み取り/
書き込み

0～5 分

Measurement
duration

6476～6477

浮動小数

読み取り/
書き込み

0.25～60 分

Validation gas
information

47238～
47253

文字列

読み取り/
書き込み

最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

Validation
concentration

47226～
47227

浮動小数

読み取り/
書き込み

0～1000000 ppmv

Validation allowance47228～
47229

浮動小数

読み取り/
書き込み

0～100%

検証開始

30015

整数

読み取り/
書き込み

0: Cancel, 1: Start

99

1=2

99
99
99
99
100
該当
1
なし

Modbus 専用のパラメータ

154

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

Performing verification サブメニュー
ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Performing verification
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Year

2495

整数

読み取り/
書き込み

9 ～ 99

Month

2494

整数

読み取り/
書き込み

0 = January
1 = February
2 = March
3 = April
4 = May
5 = June
6 = July
7 = August
8 = September
9 = October
10 = November
11 = December

100

101

Day

2493

整数

読み取り/
書き込み

1～31 d

Hour

2492

整数

読み取り/
書き込み

0～23 h

AM/PM

2496

整数

読み取り/
書き込み

0 = AM
1 = PM

Minute

2467

整数

読み取り/
書き込み

0～59 分Measurement duration

6476～6477

浮動小数

読み取り/
書き込み

0.25～60 分

Verification mode

2366

整数

読み取り/
書き込み

0 = Standard verification
3 = Extended validation
4 = Extended current output
2 = Extended validation andcurrent output

102

101
101
102
102
102

External device
information

20493～20508

文字列

読み取り/
書き込み

最大 32 文字（英字、数字、特殊文字（例：@、%、/）など）

103

Start verification

2270

整数

読み取り/
書き込み

0 = Cancel
1 = Start
1
10 = Output 1 low value
1
11 = Output 1 high value
1
12 = Output 2 low value
1
13 = Output 2 high value
18 = Prepare validation
19 = End validation

103

0～100 %

103

Progress

1

→

6797

整数

読み取り

表示項目は注文オプションまたは機器設定に応じて異なります。

Endress+Hauser155

機能説明書

J22 TDLAS ガスアナライザ

ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Performing verification
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Status

2079

整数

読み取り

0 = Failed
1 = Done
3 = Not done
8 = Busy
9 = Purging

Measured values

5512～5513

浮動小数

読み取り/
書き込み

符号付き浮動小数点数

Output values

5516～5517

浮動小数

読み取り

符号付き浮動小数点数

Measuredconcentration

36752～36753

浮動小数

読み取り

0～1000000 ppmv

Verification result

2355

整数

読み取り

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged

→

103

104
104
104

104

Verification results サブメニュー
ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Verification results
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Date/time (manuallyentered)2372～2381

文字列

読み取り

dd.mm.yy hh:mm（選択した日時
フォーマットに応じて異なります）

106

Verification ID

2315

整数

読み取り

0～65,535

106

Operating time3346～3352

文字列

読み取り

日（d）、時（h）、分（m）、秒（s）

106

Verification result

2355

整数

読み取り

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged

106

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged

106

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged

106

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged

106

SensorSensor electronic
module (ISEM)

Gas validation

156

2384

2385

5199

整数

整数

整数

読み取り

読み取り

読み取り

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Verification results
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
I/O module

System status

2386

5790

整数

整数

読み取り

読み取り

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged

106

0 = Failed
2 = Passed
3 = Not done
250 = Not supported
254 = Not plugged106

Gas validation results サブメニュー
ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Gas validation results
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Date/time (manuallyentered)

48598

文字列

読み取り

dd.mm.yy hh:mm（選択した日時
フォーマットに応じて異なります）

108

Operating time

48608～48614

文字列

読み取り

日（d）、時（h）、分（m）、秒
（s）

108

Gas validation

44668

整数

読み取り

0 = Failed
2 = Passed
3 = Not done

108

250 = Not supported
254 = Not plugged
Concentration average48034～48035

浮動小数

読み取り

0～1000000 ppmv

Concentration standard
deviation

36754～36755

浮動小数

読み取り

0～1000000 ppmv

Concentration
maximum

48229～48230

浮動小数

読み取り

0～1000000 ppmv

Concentration
minimum

48596～48597

浮動小数

読み取り

0～1000000 ppmv

109
109
109
109

Monitoring results サブメニュー
ナビゲーション：Expert → Diagnostics → Heartbeat Technology → Monitoring results
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Detector reference level4720～4721

浮動小数

読み取り

0～5 mA110

Peak 1 index delta

30581

浮動小数

読み取り

-511.0～+511.0

110

Peak 2 index delta

30672

浮動小数

読み取り

-511.0～+511.0

110Endress+Hauser

157

機能説明書6.3.8

J22 TDLAS ガスアナライザ

Simulation

ナビゲーション：Expert → Diagnostics → Simulation
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

Current input 1～nsimulation

1: 6127

整数

読み取り/
書き込み

0 = Off

Value current input 1～n

1: 6139～6140

浮動小数

読み取り/
書き込み

0～22.5 mA

Current output 1～nsimulation

1: 5939

整数

読み取り/
書き込み

0 = Off

Current output value 1～n

1: 5995～5996

浮動小数

読み取り/
書き込み

0～22.5 mA

Switch output
simulation 1～n1: 6223

整数

読み取り/
書き込み

0 = Off

Switch state 1～n

1: 6227

読み取り/
書き込み

1 = Open

読み取り/
書き込み

0 = Off

読み取り/
書き込み

1 = Open

読み取り/
書き込み

0 = Off

読み取り/
書き込み

0 = Sensor

2: 6128
2: 6141～6142
2: 5940
2: 5997～5998
2: 6224
整数

2: 6228
Relay output 1～nsimulation

1: 7523

Switch state 1～n

1: 8239

整数

2: 7524
整数

2: 8240
Device alarm simulation
Diagnostic event
category

6812
4261

整数
整数

→

110

1 = On110
111

1 = On111
111

1 = On112

6 = Closed112

1 = On112

6 = Closed

113

1 = On1 = Electronics

113

2 = Configuration
3 = Process
Diagnostic event
simulation

6.3.9

4259

整数

読み取り/
書き込み

Off
診断イベント選択リスト（選択したカテゴリに応じて異なる）

113

アクセス

選択項目/ユーザー入力/ユーザー→

Spectrum plots

ナビゲーション：Expert → Diagnostics → Spectrum plots
パラメータ

レジスタ

データ型

インターフェース
Midpoint default 1～n31090, 31092,
31094, 31096

浮動小数

読み取り/
書き込み

0～120 mA

Ramp default 1～n26750, 26752,
26754, 26756

浮動小数

読み取り/
書き込み

0～120 mA

Concentration

9455～9456

浮動小数

読み取り

0～1000000 ppmv

115

Dew point 1

21458～21459

浮動小数

読み取り

符号付き浮動小数点数116

Dew point 2

21800～21801

浮動小数

読み取り

符号付き浮動小数点数

116

Cell gas pressure25216～25217

浮動小数

読み取り

-0.5～+6.9 Bar

116

Cell gas temperature

21854～21855

浮動小数

読み取り

-20～+60 °C

116

158

114
115

Endress+Hauser

J22 TDLAS ガスアナライザ

機能説明書

ナビゲーション：Expert → Diagnostics → Spectrum plots
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー→

インターフェース
Detector reference level4720～4721

浮動小数

読み取り

0～5 mA

116

Detector zero level9667～9668

浮動小数

読み取り

0～5 mA

117

Peak 1 index

9834～9835

浮動小数

読み取り

0～511.0116

Peak 1 index delta

30581～30582

浮動小数

読み取り

-511.0～+511.0

117

Peak 2 index

27600～27601

浮動小数

読み取り

0～511.0

117

Peak 2 index delta

30672～30673

浮動小数

読み取り

-511.0～+511.0

117

Peak track index

29018～29019

浮動小数

読み取り

0～511.0

117

Peak track index delta

28814

浮動小数

読み取り

-511.0～+511.0

118

Midpoint delta

47236～47237

浮動小数

読み取り

0～120 mA

118

Analyzer control

21460

整数

読み取り/
書き込み

0 = Off

読み取り/
書き込み

0 = Off

読み取り/
書き込み

0～15

Reset
Det. 1 TIA gain

4727
29235

整数
整数

1 = On3 = Reset

118
118
119

6.3.10 SD card
ナビゲーション：Expert → Diagnostics → Spectrum plots → Chart
パラメータ

レジスタ

データ型

アクセス

選択項目/ユーザー入力/ユーザー
インターフェース

→Spectra log rate

26289～26290

浮動小数

読み取り

45～86400 s

119

Estimated number ofspectra files

24902～24903

浮動小数

読み取り

0～30

Validation log level

29082

整数

読み取り/
書き込み

0 = Off
1 = Normal
2 = Extended

119

120

255 = All
Number of validation
files

Endress+Hauser

30879

整数

読み取り

0～60

120

159

機能説明書

J22 TDLAS ガスアナライザ

www.addresses.endress.com

160

Endress+Hauser