Hipres HP1000®

High Pressure Transmitter• Pressure ranges to 5,000 bar• High pressure integrity for safe use due
to unique sensor design• Pressure diaphragm and processconnection is machined from one piece ofTitanium with no seals or welds• High resistance to overpressure andpressure transients• Silicon-on-Sapphire (SOS) sensor
technology for outstanding performanceand reliability• ATEX/IECEx option available (includes M1for mining applications) for
4-20 mA versions• DNV GL certiﬁcation available
Vers. 20/1/Eng

Hipres ® HP1000 High Pressure Transmitter
Technical Data
DescriptionThe HP1000 series extends the Silicon-onSapphire pressure sensor technology into very
high pressure applications, with operating ranges
up to 5,000 bar and still maintaining an extremely
high performance level.
The unique Silicon-on-Sapphire sensor technology provides
outstanding performance and gives excellent stability overa wide temperature range. The wetted parts and pressure
diaphragm are machined from a single piece of titanium
alloy meaning no weld joints and therefore high pressure
integrity and overload capability. All titanium pressure port
offers unbeatable corrosion resistance. With a design to meetdemanding environments, this transmitter will consistently
maintain accurate performance while sustaining high durability.

Using the industry standard autoclave process connection
enables safe and reliable sealing to such high pressures.
Available in pressure ranges from
0-400 bar to 0-5,000 bar and with electrical outputs of 0-100
mV, 0-5V dc, 0-10 Vdc and4-20 mA. Applications include aerospace, laboratory and test,
oil and gas monitoring equipment and general industrial.
An optional ATEX and IECEx approved version of this product
MWEZEMPEFPIJSVI\TPSWMSRTVSXIGXMSRJSV¾EQQEFPIKEWIW ^SRI HYWXW ^SRI ERHQMRMRKEVIEW KVSYT-1 (2:+0VYPIWJSVGPEWWM½GEXMSRSJWLMTWLMKLWTIIH PMKLXGVEJX
and DNV GL offshore standards.

Dimensions(in mm)~35
PG9
ELECTRICAL CONNECTION (mA)

Pin No.
1232 wire+supply
4-20mA signal
not ﬁtted~113to case

ELECTRICAL CONNECTION (Vdc)

Pin No.
1234 wire 3 wire
-supply common+supply +supply+output +output-output to case-output to case

1222A/F HEX

Hipres ® HP1000 High Pressure Transmitter
Technical Data
Type

HP1000/HP110

HP1xx1HP1xx2HP1003/HP1103

Silicon-on-Sapphire (SoS)

Sensor Technology:
Output Signal:

10 mV/V (4 wire)0 – 5 V (4 or 3 wire)0 – 10 V (4 or 3 wire)

4 – 20 mA (2 wire)

Supply Voltage:

10 VDC (5 – 15V)

13 – 30 VDC

13 – 30 VDC

10 – 36 VDC

Pressure Reference:

Gauge

Protection of SupplyVoltage:

Protected against supply voltage reversal up to 50 V (amplified versions)Standard Pressure Ranges(bar):

HP10xx: 0 – 600 bar; 0 – 700 bar; 0 – 1,000 bar; 0 – 1,500 bar; 0-2,000 bar. HP11xx: 0 – 2,500 bar; 0 – 4,000 bar; 0 – 5,000 bar(other ranges available)

Standard Pressure Ranges(psi):

0-10,000 psi; 0-15,000 psi; 0-20,000 psi; 0-30,000 psi; 0-40,000 psi; 0-60,000 psi; 0-72,000 psi (other ranges available)

1.5x for ranges 0 – 600 bar to 0 – 3,000 bar; 1.25x for 4,000 bar; 1.2x for 5,000 bar

Overpressure Safety:

Load Driving Capability:

Accuracy NLHR:

4 – 20 mA: RL < [UB - 10 V] / 20 mA (e.g. with supply voltage (UB) of 36 V, max. load (RL) is 1300 Ω); 10 mV/V: n/a; 0 – 5 V: max.
load RL > 5 KΩ; 0 – 10 V: max. load RL > 10 KΩ≤ ±0.25 % of span BFSL (Optional higher accuracy version of ≤ ±0.1 % of span BFSL available)

Zero Offset and Span
Tolerance:±0.5 %FS at room temperature (HP1000/HP1100: ±1 mV); ±5 %FS (approx.) adjustment with easy access trimming
potentiometers on amplified versions only

Operating AmbientTemperature:-40 °C to +85 °C (-40 °F to +185 °F)

Operating Media
Temperature:-50 °C to +125 °C (-58 °F to +257 °F)

Storage Temperature:+5 °C to +40 °C (+41 °F to +104°F) Recommended Best PracticeTemperature Effects:±1.5 %FS total error band for -20 °C to +70 °C. Typical thermal zero and span coefficients ±0.015 %FS/ °C

ATEX/IECEx Approval
Option (4-20 mA versiononly):

n/an/an/aUi = 28 V
Ii = 119 mA
Pi = 0.65 W
Li = 0.1 μH
Ci = 74 nF
Temperature Range= -20 °C to +70 °C
Max. cable length = 45 m

ATEX/IECEx Safety Values:

n/a

DNV GL Approval Class:Temperature: D; Humidity: B; Vibration: B; EMC: B; Enclosure: C (contact sales for more information)

Electromagnetic
Compatibility:

n/a

n/a

Ex II 1 G Ex ia IIC
T4 Ga (zone 0)
Ex II 1 D Ex ia IIIC
T135 °C Da (zone 20)
Ex I M 1 Ex ia I Ma (group 1M1)

Emissions: EN61000-6-3; Immunity: EN61000-6-2; Certification: CE Marked

Insulation Resistance:> 100 MΩ @ 50 VDC

Response time 10-90 %:1 mS

Wetted Parts:

Titanium alloy machined from a single piece (≥1,000 bar); Titanium alloy and SAE 316 stainless steel (<1,000 bar)Pressure Media:

All fluids compatible with Titanium alloy (≥1,000 bar); All fluids compatible with Titanium alloy and SAE 316 stainless steel(<1,000 bar)Pressure Connection:

F250-C Autoclave fitting; thread type 9/16-18UNF-2B female or M16 x 1.5 female cone seal

Electrical Connection:

Mating socket EN175301-803 Form A (ex DIN43650) rated IP65 with PG9 cable entry (other options available)

Net. Weight (Kg):

0.2 KgHipres ® HP1000 High Pressure Transmitter
Order Matrix

Output

10 mV/V

0-5 V

0-10 V

4-20 mA

Sensor Range

WiresType

Model to 2,000 bar (incl. 30,000 psi)

4HP1000Model above 2,000 bar

4HP1100Model to 2,000 bar (incl. 30,000 psi)

4HP1001Model above 2,000 bar

4HP1101Model to 2,000 bar (incl. 30,000 psi)

3

HP1011Model above 2,000 bar

3HP1111Model to 2,000 bar (incl. 30,000 psi)

4HP1002Model above 2,000 bar

4HP1102Model to 2,000 bar (incl. 30,000 psi)

3

HP1012Model above 2,000 bar

3HP1112Model to 2,000 bar (incl. 30,000 psi)

2HP1003

Model above 2,000 bar

2HP1103

ElectricalConnection/Options

Pressure Range

ProcessConnection

Electrical Connection/Options
DIN EN175301 plug and socket-

Cable outlet 1m screenedA

M12 connector

B

Cable outlet 1m screened IP67 protection

CATEX/ IECEx certified with DIN EN175301 plug and socketEX

DNV GL Approval

M

DNV GL Approval plus ATEX/IECEx certified

MEX

Pressure Range in bar
0-600 bar600

0-1000 bar

1000

0-1,500 bar

1500

0-2,000 bar

2000

0-3,000 bar

3000

0-4,000 bar

4000

0-5,000 bar

5000

Process Connection
Autoclave F-250-C femaleDE

M16 x 1.5 female cone seal

FK

Order Number Example

HP1100A1000DE

For options not listed please contact the sales teamDISCLAIMER : ESI Technology Ltd operates a policy of continuous product development. We reserve the right to change specification without
prior notice.All products manufactured by ESI Technology Ltd are calibrated using precision calibration equipment, traceable to national
measurement standards.

t. +44(0)1978 262 255
e. sales@esi-tec.com

www.esi-tec.com