ORDERING INFORMATIONMODEL : M2EXR
FACTORY USE ONLY➠➠➠➠➠➠➠➠

PLEASE FILL IN THIS SECTION
ModelJob No.

Approved by(Sales office)Company

Ser No.

Issued by(Sales office)

Name

Sales

Approved by(Factory)
Set by(Factory)P/O No.

Ser No.
Specify the items you want to change.
Default setting will be used if not specified.
■ INPUT SETTING
ITEM

Input range

Temperature unit

SET VALUE□ JPt100（JIS’89）□ Pt100（JIS’89）□ Pt100（JIS’97, IEC）
□ Pt50Ω（JIS’81）□ Ni508.4Ω
□ Pt1000□ Ni100□ Cu10 @ 25℃□ Celsius□ Fahrenheit□ Absolute

0 % input setting

COMMENTS

Factory
Internalcheck

Pt100（JIS’97, IEC）

Choose among 8 types□

Celsius

Choose among 3 types□

Specify within settable range in the table 1.

□

0.0

100 % input setting

Input decimal point

DEFAULT VALUE

100.0□ No decimal point□ 1 place of decimals□ 2 places of decimals

Choose among 3 types
1 place of decimals

Set the decimal point position for 0 %□input setting and 100 % input setting.

Sensor wires□ 2 wires□ 3 wires3 wires

Choose among 2 types□

Burnout□ None□ Down□ UpUP

Choose among 3 types□

0 sec.

Specify the range between 0 and 30 sec□

Filter time constant *1

MG CO., LTD.
www.mgco.jp

M2EXR ORDERING INFORMATION ESU-5144 Rev.5 P1/4

■ OUTPUT SETTING
ITEM

Output range

SET VALUE□ 0 to 20 mA DC□ -5 to +5 V DC
□ -10 to +10 V DC

DEFAULT VALUE

0 to 20 mA DC

0 % output setting

4.000 mA
-5.000 V-10.000 V

100 % output setting

20.000 mA
5.000 V
10.000 V

COMMENTS

FactoryInternalcheck

Choose among 3 types□

Specify within settable range in the table 2.

□COMMENTS

Factory
Internalcheck■ DISPLAY SETTINGITEM

SET VALUEDEFAULT VALUE

0 % output scaling

0.00

Specify the range between –99999 and999999.

□

100 % output scaling

100.00Specify the range between –99999 and999999.

□

2 places of decimals

Choose among 6 types
Set the arbitrary decimal point position for
0 % output scaling and 100 % output
scaling.

□%

Choose from the table 3, or specify fromtable 4 ‘Settable characters’ within 13characters.

□Upper: INPUT
Lower: PERCENT

Choose from the setting value in the table
5.

□Brightness

4

Specify among 1 (darkest) to 4 (brightest).

□Display timeout

10 min.

Specify the range from 0, 1 to 60 min.
Set ‘0 min’ to display ‘always on’.

□Output decimal point□ No decimal point□ 1 place of decimals□ 2 places of decimals□ 3 places of decimals□ 4 places of decimals□ 5 places of decimals

Unit (OUT Scaling)

Display setting

Upper:Lower:*1. Filter time constantSet filter time constant of the first order lag filter. The first order lag filter is available with setting time. When ‘0’ is set to this parameter, the firstorder lag filter is not available (Response time: ≤ 0.5 sec. (0 ➝ 90 %)).

The first order lag filter is equivalent to general CR filter. The setting

time constant is the time to follow until about 63 %, when input varies from 0 % to 100 %.

MG CO., LTD.
www.mgco.jp

M2EXR ORDERING INFORMATION ESU-5144 Rev.5 P2/4

Table 1
MIN. SPAN
INPUT RANGE

C, K

SETTABLE RANGE
CFK

JPt100 （JIS’89）-200.00 to +500.00-328.00 to +932.00

73.15 to 773.15

Pt100 （JIS’89）-200.00 to +650.00-328.00 to +1202.00

73.15 to 923.15

Pt100 （JIS’97, IEC）

-200.00 to +850.00-328.00 to +1562.00

73.15 to 1123.15-200.00 to +649.00-328.00 to +1200.20

73.15 to 922.15

Pt50Ω （JIS’81）

20.00

Ni508.4Ω

°Ｆ

36.00

-50.00 to +200.00-58.00 to +392.00

223.15 to 473.15

Pt1000-200.00 to +850.00-328.00 to +1562.00

73.15 to 1123.15

Ni100-50.00 to +200.00-58.00 to +392.00

223.15 to 473.15

Cu10 @ 25℃-50.00 to +250.00-58.00 to +482.00

223.15 to 523.15

Note: For set value, specify so that value is within the settable range depending on type of the RTD.

Table 2OUTPUT RANGECurrent output
Voltage output

MIN. SPANSETTABLE RANGE

0 to 20 mA DC

1.000 mA

0.000 to 20.000 mA-5 to +5 mV DC

0.250 V

-5.000 to +5.000 V-10 to +10 V DC

1.000 V

-10.000 to +10.000 V

Table 3AVAILABLE UNITS
DC, AC, mV, V, kV, μA, mA, A, kA, mW, W, kW, var, kvar, Mvar, VA, Hz, Ω, kΩ, MΩ,cm, mm, m, m/sec, mm/min, cm/min, m/min, m/h, m/s2, inch, L,L/s, L/min, L/h, m3, m3/sec, m3/min,m3/h, Nm3/h, N∙m, N/m2, g, kg, kg/h, N, kN, Pa, kPa, Mpa, t, t/h,°C, °F, K, %RH, J, kJ, MJ, rpm, sec, min, min-1, pH, %, ppm, deg, (blank),Table 4
SETTABLE CHARACTERS
0 − 9 A − Z a − z ! ” # $ % &’ ( ) = - +  ^ |@`[]{};:<>?_,./MG CO., LTD.
www.mgco.jp

M2EXR ORDERING INFORMATION ESU-5144 Rev.5 P3/4

Table 5Upper
SETTING VALUEDESCRIPTION

INPUT

Input engineering unit value

RESISTANCE

Input resistance

PERCENTPercent value*2

OUTPUT

Output engineering unit value

OUTPUT (Scaling)

Output scaling

Lower
SETTING VALUEDESCRIPTION

INPUT

Input engineering unit value

RESISTANCE

Input resistance

PERCENTPercent value *2

OUTPUT

Output engineering unit value

OUTPUT (Scaling)

Output scaling

None

No display*2. Input percent value.

MG CO., LTD.
www.mgco.jp

M2EXR ORDERING INFORMATION ESU-5144 Rev.5 P4/4