2065FTPEX(E)-v1.2.0

Ethernet Communication Software
FTPExtractor
Operation Manual

KIMOTO ELECTRIC CO., LTD.

2065FTPEX(E)-v1.2.0

Abstract
We are delighted that you have decided to use KIMOTO’s analyzers. FTPExtractor is a free softwarethat provides your PC the function to communicate with all our analyzers that have Ethernet port. You
can download the data stored in the instrument to your PC, using this software. In addition, you are alsoable to download it from PC-card (CompactFlashTM, 600 series (optional)) or an internal storage (700series). Ethernet-HUB supported 10/100 BASE-T should be necessary to communicate. Microsoft ®
Windows-based PC (7, Vista™, XP) is needed to operate.

Contents
1. INSTALLATION
2. NETWORK CONNECTION
2.1 Ethernet setting of PC
2.2 Connection method between KIMOTO’s analyzer and PC
2.2.1 Connection using a cross cable
2.2.2 Connection using a Ethernet HUB
3. Operation
3.1 Connection
3.2 Data Downloading4. APPENDIX
4.1 Major IP Address of our instruments
4.2 Typical file name of measured data
4.3 Version of FTP Extractor
4.4 Download Site of FTP Extractor

12222344566667

2065FTPEX(E)-v1.2.0

1. INSTALLATION
Start the installation process using install program "FTPExtractor2.exe". You can download the installer on our website (Refer Sec.5.4)
Wizard screen appears when you open the install program. Install following the instructions on thescreen.
After installation was performed, you will be able to find FTPExtractor to boot on the desktop as below(from "Start" button -> "Programs" -> "Kimoto Utils" -> "FTPExtractor").

-1-2065FTPEX(E)-v1.2.02. NETWORK CONNECTION
All KIMOTO’s analyzers have an Ethernet port. You can easily download the data to your PC via
Ethernet. It may needed for some networking knowledge such as administrative privileges of Windows
in order to setup. If you have any trouble to setup, .please ask to a network administrator or an authority
on networking.
2.1 Ethernet setting of PC
It is necessary to setup Ethernet connection of your PC before data communication. All our analyzers
manufactured after December 2005 have dynamic host configuration protocol (DHCP) that assign IPaddresses automatically. For other analyzer manufactured before December 2005, it is need to setup theIP address and subnet mask manually.
For example (manual setting);
IP address:
192. 168. 0. 10
Subnet mask: 255. 255. 255. 0
2.2 Connection method between KIMOTO’s analyzer and PC
There are two methods for communication between our analyzer and PC. One is one-to-one
communication using a cross-cable Another is multiaccess communications using Ethernet HUB.
2.2.1 Connection using a cross cable
The following figure shows a connection using a crossover Ethernet cable. Specification of the PC, it
may not communicate infrequently. Then, you had better use HUB as section 2.2.2.

PC

SPM-613KIMOTO

10/100 Base-T Cross Cable
10/100 Base-T Cross Cable
1 RD+
2 RD3 TD+456 TD7812341 RD+2 RD3 TD+456 TD785678UpLink

HUB

10/100 Base-T Straight Cable

PC-2-SPM-613

KIMOTOKIMOTO10/100 Base-T Cross Cable
1 RD+
2 RD3 TD+456 TD7
81 RD+
2 RD3 TD+456 TD782065FTPEX(E)-v1.2.0

2.2.2 Connection using a Ethernet HUB
The following figure shows a connection using Ethernet HUB and some straight cables. In the case ofthe connection using a HUB, it allows the multiple communications between your PC and our analyzers
simultaneously.

12345678UpLink

HUB

10/100 Base-T Straight CablePC

SPM-613KIMOTO

10/100 Base-T Straight Cable
1 RD+
2 RD3 TD+456 TD7
81 RD+
2 RD3 TD+456 TD78-3-KIMOTO

2065FTPEX(E)-v1.2.0

3. Operation
FTPExtractor is the software in order to downloading the data stored in internal storage (600/700series) and PC-card (optional memory card for 600series) to your PC via network. It is also available
to download the message (operational history)
files. Please boot FTPExtractor after setting up thenetwork.
3.1 Connection
1. Boot “FTPExtractor.exe” then main screen
appears. Click “Dir List”, then “Target IP
address” window opens. Pull-down menu indicated “Candidate” is also available, if youranalyzer is included the list.
2. Push “OK”, after IP address input.
3. If the line is clear, “STORAGE” will be indicated in the list. In STORAGE, the data are
stored in the internal storage (600/700 series)
and PC-card (600 series). If your analyzer
does not have PC-card slot, it is not indicatedthe data in PC-card.
[!] If your OS is Windows 7 or Windows Vista,“Windows Security Alert” would be indicatedwhen you click “OK” button. In order to allow
communication with your analyzer, check-off
all boxes and click “Allow Access” button.
Also, administrative privileges are required toallow access.

-4-2065FTPEX(E)-v1.2.0

3.2 Data Downloading
4. Click “Save file” button on the main screen
after selecting the file you want to save. Thena confirmation dialog displays. Please specify
the destination. Downloading starts when youclick “OK”. If the data exists other level, FTP
Extractor shows “ △ ” at the end of the folder
name. When double-clicking this item, “ ▲ ”
will be displayed. Then you can see the data
including each level. You can download thedata stored in different level. Also, you candownload a whole data in each directory. Inthis case, click “Save File” after selecting adirectory you want to download.
5. To quit FTPExtractor, click “Exit” button on
main screen.

-5-2065FTPEX(E)-v1.2.0

4. APPENDIX
4.1 Major IP Address of our instruments
TYPE

IP ADDRESS

Data Logger

192.168.0.110

Nitrogen Oxides Analyzer

192.168.0.120

Ozone Analyzer

192.168.0.130

Sulfur Dioxide Analyzer

192.168.0.140

Particurate Matter Monitor

192.168.0.150

4.2 Typical file name of measured data
TYPICAL DATAINFORMATION

MSzzzhhh.TXT

Message File

HRzzzhhh.CSV

1 hour value file

DFzzzhhh.TXT

DA set value fileSFzzzhhh.TXT

System set value fileAFzzzhhh.TXT

Application set value file"zzz" means the zone number of the IP address, while "hhh" means the host number. For example, if
the IP address is [192.268.0.110], the name of "Message" file becomes "MS000110.TXT".
4.3 Version of FTP Extractor
You can see the version number of FTPExtractor with "Version" screen. Push "Version" button on themain screen to open it.

-6-2065FTPEX(E)-v1.2.0

4.4 Download Site of FTPExtractor
The newest installer is located on the internal web site at;		

http://www.kimoto-electric.co.jp/support/index.html

This manual explains for the following versions of software.		

PID

2065

Ver. 2.0.6 or later-7-2065FTPEX(E)-v1.2.0

KIMOTO ELECTRIC CO., LTD.
3-1 Funahashi-cho Tennouji-ku Osaka 543-0024 JAPANTEL:+81-6-6768-3401
FAX:+81-6-6764-7040
URL:
http://www.kimoto-electric.co.jp/E-mail: sales@kimoto-electric.co.jp