Contents
p. 2Applic.
pp. 4-8

Selection
pp. 9-11

Products
pp. 12-87

GenRadproducts
pp. 50-87

精密手動またはプログラマブル
Precision Manual or電圧及び電流源
Programmable Voltage andCurrent Source

Index
p. 89

VI-700 Series
多くの機能と豊富なオプションを備えたVI-700
Various features and options make the VI-700
は手動あるいはプログラマブル電圧及び電流源
an effective choice for many manual and proとして多くの応用で非常に役立つ装置です。
grammable voltage and current applications.
Voltage output rangeof 100 μV to 20 V or 200 V
出力電圧範囲：100μV出20Vあるいは200V

snoaticcifiepdsevor
出力電流範囲：0.1μA出20mAあるいは200mACurrentoutput rangeof 0.1 μA to 20 mA or p
200m
ImA
デジット分解能:
0.75LSD
3½
digit resolution,3・1/2、確度:
0.75 LSD accuracy

入力はロータリスイッチの手動操作あるいはIEEE-488
Manualthumbwheel operation or programmable又はパラレルBCDインターフェイスからのデジタル入力
digitalinput, IEEE-488 or parallel BCD interface.
オプションの携帯用パワーパック
Optional portable power packMay be used as a component in a closed loop
クローズドループのコントロールシステムコンポーネント
control systemとして使用可

Model VI-700 Precision Voltage and Current Source

仕 様
SPECIFICATIONS
Voltage
電圧
Output
出力範囲RangesResolution
分解能Accuracy
確度

OutputCurrent
出力電流Output
出力インピーダンス
Impedance
Noise andノイズ及びRipple
リップル
Tempco
温度係数

200 mV, 2 V, 20 V;(200 V with HV option).
100 μV, 1 mV, 10 mV;(100 mV with HV option).
±(75 ppm + 50 μV); 200 mV range;±(75 ppm + 0.25 mV), 2 V range;±(75 ppm + 2.5 mV), 20 V range;
at 23°C, no load, after warmup.
70 mA, (200 mA with HP option) maximum
load up to 20 V; (10 mA for 200 V range withHV option); with LED overload indicator.

5 mΩ typical, 40 mΩ for 20 V range.
50 μVrms on 200 mV and 2 V ranges;50 μVrms on 200 mV and 2 V ranges;
100 μVrms on 20 V range.
±10 ppm/°C typical;±15 ppm/°C max.

Digital Input
デジタル入力

一般General
31/2 digit parallel BCD or IEEE-488
interface; a front panel push-button
selects REMOTE or LOCAL operation.

Isolation
隔離

Floating outputs, optically isolated fromdigital inputs; voltage and current outputs may be used in either polarity.

Power
電源

105-125 V or 210-250 V;
50-60 Hz; 5 W.
21.6 cm W x 12.6 cm H x 22.9 cm D
(8.5" x 2.8" x 9.0" ) for Standard Model;
21.6 cm W x 8.7 cm Hx 23.5 cm (8.5"x 3.44" x 9.25" ) for IEEE-488 interface
option.
2.3 kg (5 lb).

Requirements
Dimensions
大きさWeight
重さ

運転温度範囲：Operating Temperature Range: 10°C to 40°C.

電流
Current
出力範囲
OutputRanges200 μA, 2 mA, 20 mA,
200 mA (with HI option).
Resolution
0.1 μA, 1 μA, 10 μA;
分解能
100 μA (with HI option).
Accuracy±(75 ppm±(75ppm+ .05 μA); +.05μA);
200 μA; 200μA
確度±(75 ppm + 0.25 μA), 2 mA;±(75 ppm + 2.5 μA), 20 mA; 23°C, no load.
Compliance0-20 V; with LED overload indicator.
適合電圧
VoltageNoise8-20 nA.
ノイズ

保管温度範囲：StorageTemperature
Range: -40°C to 60°C.

ORDERING INFORMATION
オーダー情報Model VI-700-110
Model VI-700-220-HP Option-HI Option-HV Option-RS Option-BCD Option-IEEE Option
IEEE-488
IEEE-488
BP-511

IET LABS, INC. in the GenRad Tradition
534 Main Street, Westbury, NY 11590
IET cat/VI-700, Rev J3/01-17-05

Precision Voltage & Current Source; 110 Vac op.
Precision Voltage & Current Source; 220 Vac op.
High power output, up to 200 mA output
200 mA output (in addition to 200 μA-20 mA)
200 V output (in addition to 100 μV -10 mV)
Remote sensing directly to load
Parallel BCD Remote Programming
IEEE-488 Remote Programming
Interface hardware (see p. 48)
Interface software (see pp. 46-47, 49)Battery Pack, AC source, 115 V, 60 Hz, 300 W(see p. 48)

www.ietlabs.com
TEL: (516) 334-5959 • (800) 899-8438 • FAX: (516) 334-5988
41