Absolute and Gauge Pressure Transducers

Series 2000 Absolute Pressure Transducer

Series 2000

Series 2000 Gauge Pressure Transducer

10 ABSOLUTE RANGES
7-16 psia (0.05-0.11 MPa)
0-15 psia (0.10 MPa)
0-23 psia (0.16 MPa)0-30 psia (0.21 MPa)0-45 psia (0.31 MPa)
0-100 psia (0.69 MPa)
0-200 psia (1.38 MPa)
0-300 psia (2.07 MPa)
0-400 psia (2.76 MPa)
0-500 psia (3.65 MPa)FEATURES & PERFORMANCE**
Frequency Outputs
0.01% Typical Accuracy
Parts-per-billion Resolution*
Low Power Consumption
High Stability and Reliability
On-Board Coefficient Storage
On-Board Application Memory
Fully Calibrated and Characterized
NIST Traceable Calibration - CE Compliant

7 GAUGE RANGES± 2 psig (±0.015 MPa)
0-15 psig (0.10 MPa)0-22 psig (0.15 MPa)0-30 psig (0.21 MPa)
0-100 psig (0.69 MPa)
0-150 psig (1.03 MPa)
0-200 psig (1.38 MPa)

APPLICATION AREAS
Metrology
Hydrology
Aerospace
Meteorology
Oceanography*With Digiquartz® Nano-resolution Electronics**Products defined by specification control drawing

Paroscientific manufactures and sells a complete line of high precision pressure instrumentation. Parts-per-billion resolution
and typical accuracy of 0.01% are achieved even under harsh environmental conditions. Digiquartz® Transducers areessential for a variety of application areas where high resolution, accuracy, reliability, ruggedness, long-term stability,
low power consumption and low cost of ownership are important requirements.
The remarkable performance of these transducers is achieved through the use of a precision quartz crystal resonator whose
frequency of oscillation varies with pressure induced stress. A quartz crystal temperature signal is provided to thermallycompensate the calculated pressure and achieve high accuracy over a broad range of temperatures. The transducers include
integral shock protection to withstand extremely high acceleration, shock, and vibrational loads.
Our Quality Management System is certified to the requirements of the ISO 9001 International Quality Standard. It provides
consistency in our products and processes from design and development through production, calibration, test, and servicing.
Our quality system and commitment to excellence ensure customers of outstanding products and services. We offer a marketleading five year limited warranty on all Digiquartz® Transducers with the first two years covered at 100%.

Paroscientific, Inc.
Digiquartz® Pressure Instrumentation

Absolute Pressure Transducers

Series3000

AbsolutePressure

SeriesSeries 3000 & 4000

4000

AbsolutePressure

10 ABSOLUTE RANGES
0-1000 psia (6.89 MPa)
0-2000 psia (13.8 MPa)
0-3000 psia (20.7 MPa)0-4400 psia (30.3 MPa)
0-6000 psia (41.4 MPa)
0-10,000 psia (68.9 MPa)
0-15,000 psia (103 MPa)
0-20,000 psia (138 MPa)
0-30,000 psia (207 MPa)
0-40,000 psia (276 MPa)FEATURES & PERFORMANCE**Frequency Outputs
0.01% Typical Accuracy
Parts-per-billion Resolution*
Low Power Consumption
High Stability and Reliability
On-Board Coefficient Storage
On-Board Application Memory
Fully Calibrated and Characterized
NIST Traceable Calibration - CE Compliant

APPLICATION AREAS
Metrology
Oceanography
Energy Exploration
Portable Standards
Process Control Systems*With Digiquartz® Nano-resolution Electronics**Products defined by specification control drawing

Paroscientific manufactures and sells a complete line of high precision pressure instrumentation. Parts-per-billion resolution
and typical accuracy of 0.01% are achieved even under harsh environmental conditions. Digiquartz® Transducers are
essential for a variety of application areas where high resolution, accuracy, reliability, ruggedness, long-term stability,
low power consumption and low cost of ownership are important requirements.
The remarkable performance of these transducers is achieved through the use of a precision quartz crystal resonator
whose frequency of oscillation varies with pressure induced stress. A quartz crystal temperature signal is provided tothermally compensate the calculated pressure and achieve high accuracy over a broad range of temperatures. Thetransducers include integral shock protection to withstand extremely high acceleration, shock, and vibrational loads.
Our Quality Management System is certified to the requirements of the ISO 9001 International Quality Standard. It provides
consistency in our products and processes from design and development through production, calibration, test, andservicing. Our quality system and commitment to excellence ensure customers of outstanding products and services. We
offer a market-leading five year limited warranty on all Digiquartz® Transducers with the first two years covered at 100%.