45

Core Technology
With 45 years of experience in the optics industry Santec has command of the key manufacturing processes to build lasers,
instruments and components that meet the stringent demands for excellence by our customers. By committing to innovation as
well as listening to our customer requests, Santec has succeeded in introducing solutions for a broad range of new opticalapplications. Our drive for Optopia has resulted in steps that minimize the timeline for a product to go from conception tocommercial applications.

43

INDEX
Developing and manufacturing tunable lasers, optical
instruments, fiber-optic components and laser source for
optical coherence tomography.

Mototaka Tei

IT Solution business in Japan.

Executive Vice President and CSO

Service for customers in more than 30 countries, which include
most of the world’s major telecommunications companies andtelecommunications system manufactures.

The Photonics Pioneer
PRODUCT LINE-UP
Core Technology
Medical Applications
Quality Products for Fiber Optic Applications
ISO Certifications
FACT SHEET
Test & Measurement

and CTO

4

Optical Test Platforms

5

Component Testing

7

IL/RL Meters

9

Optical Switches

11

Optical Attenuator

13

Polarity Testers

14

Tunable Filters

15

Launch Conditioning

16

Swept-Wavelength Component Test System
Optical Power Meter
Polarization Control Unit

Founded on Aug 25, 1979 by Dr. Masao Sadamura
Listed on the Standard market
in Tokyo Stock Exchange.

Swept Photonics Analyzer
Polarization Extinction Ratio Meter
Polarization Dependent Loss Meter

Santec Holdings Corporation
Santec AOC

Santec LIS

Santec OISSantec Japan

Corporation

Corporation

Corporation

Corporation
SantecJimbocho Office, TokyoFY 2021

15,246

Light Sources

Polarized Light Source
Stabilized Light Source

293 employees in the world. (As of March 31, 2023)

FY 20221

High Performance Tunable Laser
Wavelength Selectable Laserand COO

Corporate Headquarters in Japan and five subsidiaries in theworld.

Bench Top Tunable Lasers

8,890

3,001

1,647

4,978

4,978

Santec USA

SantecCorporation

Europe Ltd.

Santec Photonics
Laboratory
Silicon Valley

Santec(Shanghai)Co., Ltd.

SantecCalifornia
Corporation

SantecCanadaCorporation

Beijing Office
China
ShenzhenSupportCenterOptical Return Loss Meter
Backreflection Meter
Insertion Loss Meter
Discontinuity Test System
Optical Switch
OEM Optical Switch
Multichannel Optical Power Meter
High Performance Optical Power Meter
Optical Variable Attenator
Optical Backreflector
Polarity Test Meter
Remote-Head Polarity Detector
Optical Variable Attenator
Launch Condition Meter
Mode Fill Conditioner

Components
Spatial Light Modulators

17

Programmagle Optical Filters

21

Optical Monitors

22

Variable Attenuator

23

Optical Filters

24

Medical Applications

25

Liquid Crystal Based Spatial Light Modulator

USA

400 Kelby Street Suite 1501, Fort Lee, NJ 07024, USA
Toll Free: +1-800-726-8321 (santec-1)
Tel: +1-201-488-5505 Fax: +1-201-488-7702

Europe99 Park Drive, Milton Park, Abingdon
Oxfordshire, OX14 4RY, UK
Tel: +44-20-3176-1550

Programmable Arbitrary Spectral Shaping
Multichannel Optical Power Monitor
Ultimate Tap Photodetector
MEMS Attenuator

MEMS Tunable Filter
Advanced Optical Biometer

Optical Coherence Tomography
27

OCT Scanning Lasers

29

OCT Components

31

3D Optical Profiling
Wafer Thickness Mapping

33
36
39

SS-OCT system for research, feasibility studies and product development
Features of OCT

CANADACANADACALIFORNIAOCT Applications

160 Michael Cowpland Dr Kanata, ON K2M 1P6 Canada

CALIFORMA

4750 Calle Quetzal Camarillo, CA 93012 USA

Czech Office (Czech)

Tunable VCSELMEMS based HSL
Polygon scanner based HSL
Compact and High-performance SS-OCT Engine
Features of IVS software
OCT-grade Balanced Detector
Integrated Interferometer Module
High Speed Imaging DAQ-Board for HSL

Business Solutions

LiveSupporter / Online Screen View / Online Screen Assist / AppCheck

High Performance Tunable Laser

Lase rTSL-770 (Flagship model/ 7th generation)

●
●
●
●
●
●

L a s erTSL-570 (High performance model)
Santec’s all-new TSL-570 is a high performance tunable
laser with a wide tuning range, high power output and highsignal-to-noise ratio. It uses a new optical cavity design
with pre cise spe e d control up to 20 0 nm /s andsub-picometer resolution and accuracy.
The TSL-570 is a high specification, full feature instrument
suitable for all applications. It has a simple to use touch
panel display as well as Ethernet, GPIB and USB interfaces
for remote control. A Wake-on-LAN (WoL) feature provides
convenience for remote installations. Models are available
to cover from 1240 to 1680 nm with output powers up to 20
mW and feature 0.1 pm resolution, sub-pm accuracy and am a r k e t l e a d i n g, 9 0 d B / 0 .1 n m, u l t r a - l o w l e v e l o f
spontaneous source emission. Santec’s new sealed laser
cavity is mode-hop-free and provides a stable output at
every wavelength.

Wide tuning range from 1480-1640 nm
Mode-hop-free with rapid sweeps up to 200 nm/s
High output power +13 dBm
High signal-to-noise ratio 90dB/0.1 nm
High wavelength accuracy 0.3 pm(typ.)Narrow linewidth <60 kHz

Features●●●●< TSL series performance comparison >●Fast, up to 200 nm/s, wavelength sweeps
Wide tuning range lineup: 1240 to 1680 nm
Wavelength resolution: 0.1 pm
High Signal-to-Noise Ratio: 90 dB/0.1 nm
High output power +13 dBm (+20 dBm available, Type H)

●
●
●
●
●

Optical component characterization
Photonic integrated circuit testing
Quantum photonics
Optical spectroscopy
Interferometry

Typical Specifications

Type P
Type H

(pm)

(dBm)

200

> 13

+/- 1

> 13

1240 - 1640 nm

(kHz)

200

+/- 5

200

1260 - 1360 nm
1500 - 1630 nm

+/- 2

> 20

100

TSL-570

1560
1490

TSL-570 / TSL-770
TSL-770

1630

1500

TSL-570

1630

1480

TSL-570
TSL-570

1355
1355
1240

TSL-570
TSL-570

0
-20
-40
-60

-80

1505

E-band1360

1680

S-band
1460

C
1530

L-band1565

U-band
1625

1570

-20
-40

-60
-80
1240

1280

1675

TSL-770は、第7世代となる高性能・高速波長可変レーザです。レーザ共振器および制御回路を完全に新設計することで、従来比2倍の高速スキャンが可能となり、かつ狭スペクトル線幅と低ノイズ化を実現しました。本製品は最大160nmの広帯域波長可変が可能で、測定のスループット、精度を向上させることができます。これにより評価現場、R&D、基礎研究など様々なアプリケーションで理想的なツールとなっております。また、光学系の振動対策、制御系のノイズ低減により外部共振器型波長可変光源の特長である非常に高いQ値特性を活かした狭スペクトル線幅と、高周波数安定度の両立を実現したことにより、シリコンフォトニクス等、非常に高い周波数精度を求められる最先端の研究分野でもご利用いただけます。

1360

TSL-570

0
-10
-20

99.75

100
100.25Frequency [MHz]1400

Wavelength stability6

TSL-770

-30
99.5

1320

Wavelength [nm]Line width (Delayed Self-Heterodyne)

1380

O-band1550
1560
Wavelength [nm]

0

10

1485

Full-band TSL

1540

Wide tuning range (O-band)

20

1680

1640

1360

1240

1260

High Power and Signal-to-noise ratio

-100
1530

Wavelength range

1260

20

60

+/- 201240 - 1680 nm

Line width

100.5

WL error
Temp.

4

40
35

2

30

0

25

-2

20

-4

15

-6

0

4

8

12
Time [hrs]1620

Temperature [degreeC]

TSL-5701480 - 1640 nm

Wavelength accuracy

Output Power(nm/s)

Output power [dBm]

Type C

Sweep speed

W avelenght error [pm]Type A(nm)

Output power [dBm]

TSL-770

Wavelength range

Spectrum density [dBm]

Product10

TSL-570は、高速スキャンと高い安定性の両方を備えた波長可変レーザーです。新しい光共振器設計により最大200nm/sの掃引速度制御とサブピコメートルの分解能と精度を備えています。 さらに、新しい密閉型レーザーキャビティによりモードホップフリーで、すべての波長で安定した出力を提供します。また、使いやすいタッチパネル
ディスプレイと、リモートコントロール用のイーサネット、GPIB、USBインターフェースを備えており、操作性も向上しています。ゆえに、光通信用光学部品の特性評価、フォトニック集積回路や量子フォトニクス等の研究開発など様々なアプリケーションで理想的なツールとなっております。

Light Sources
Full-band TSL

Lase r

The Full-band TSL is a full-band, high performance tunable
laser system that covers the ultra-wide tuning range of 1240
to 1680nm (O-band to U-band). The system combines up to
four Santec’s tunable lasers (TSL-570 and TSL-770) with an
optical switch module (OSU-110) and control software. The
Full-band TSL can be easily controlled via GPIB and USB on
PC using the custom GUI.

Features●●●●

Wide wavelength range 1240-1680 nm
Mode-hop-free wavelength sweeps
200 nm/s sweep speed
Easy and automatic operation by exclusive

The PLS-100 is a light source with well defined polarization.
It features stable, high output power while maintaining ahigh polarization extinction ratio (PER) of 45 dB (typ.).
Combining the PLS-100 and PEM-340 (Polarization
Extinction Ratio Meter) provides an optimized system for
PER measurement of polarization maintaining (PM) fiber
optic devices in both R&D and production environments.

Features●●●●

Large polarization extinction ratio, up to 45 dB (typ.)High output power >0dBm
Excellent power stability <0.1dB/h
Wavelength range 1310 ± 20 nm or 1550 ± 20 nm

Applications●●

Alignment of polarization axis for optical devices
Connectorized PMF patchcord evaluation

OSU-110The OSU-110 is an optical switch that covers a wide bandof 1240-1680 nm. By combining the OSU-110 and multiple
santec TSLs, it is possible to construct a tunable lightsource system with a ultra-wide band range. The OSU-110
is one of the key parts of test system automation and are
ideal for reducing measurement time and improving
measurement reproducibility.

Features●●●

Wide wavelength range 1240-1680 nm
SMF or PMF
Fast switching speed

WSL-110
The WSL-110 is a compact and cost effective tunable laser
source for use in WDM systems and optical measurement
applications. This laser can be tuned to any wavelength withhigh, 1 MHz resolution in both C-band or L-band, covering a
38nm tuning range. An internal wavelength locker provides
high wavelength accuracy and stability.

Features
●
●
●
●
●
●

C-band or L-band tuning with high resolution
Fine tuning available with 1 MHz resolution
Narrow linewidth <100 kHz
High output power >+15 dBm
Integrated wavelength locker
Single or two ports available

Full-band TSLは、1240-1680nm(O-bandからU-band)の広帯域をカバーする波長可変レーザー光源システムです。最大4台の波長可変レーザーTSL-570と光スイッチ
OSU-110で構成されており、コストパフォーマンスが高い光源システムとなります。
OSU-110は、1240-1680nmの広帯域をカバーする光スイッチです。波長可変レーザーTSLシリーズと組み合わせることによって広帯域をカバーする波長可変レーザー光源シ
ステムFull-band TSLとしてもご利用いただけます。
WSL-110は、1MHzの高分解能で任意の波長が設定可能な波長可変光源です。内部の波長ロッカーにより高波長精度と高波長安定度を実現しています。

Stabilized Light Source
SLS-100
The SLS-1x x series of fixed light sources is a highly
configurable system with many different options for lightsources. There are three different versions of the SLS
series. The SLS-100 is a fixed bank of light sources, theSLS-110 is a switched optical source, and the SLS-120 is a
fixed bank of light sources dedicated to CWDM DFB laser
sources. Each of these systems come in either a half rackor full rack 2U enclosure.
The SLS-100 can be configured with up to 24 discrete LEDsor lasers (1mW type) or the system can be mixed andmatched with different source types, wavelengths, or fiber
types to accommodate various applications in one box.
The SLS-110 is a system that has an internal source orsources that can be combined and switched at the output.
This is convenient if one needs combined wavelengths outof many port, such as 24 channels with 1310,1490,1550,1625,
which would not be possible in the SLS-100 series.
For dedicated CWDM applications, the SLS-120, canaccommodate all ITU-T G.694.2 channels (1270 to 1610,
20nm apart) in one box with standard 1mW DFB sources.
Also available with 10mW or 20mW DFB lasers with at least
30dB of side mode suppression ratio.

Features
Up to 24 LEDs or Lasers in one rack
Factory configurable wavelength mix including various SM
and MM source types
Adjust power levels
Customer definable launch condition
Combined sources in one port (850/1300 MM, 1310/1550 SM)
Support common connector types on front panel (FC, SC, ST)
Always on optical source
●
●

●
●
●
●
●

Applications●●●

Optical alignment
Environmental testing of IL for cables and optical components
Real time IL monitoring

PLS-100は、 45dBの消光比を維持しながら高出力および高出力安定度を実現した偏光光源です。PEM-340と組合せることにより、PER測定が可能となります。
SLS-100は、最大24個の個別LEDレーザー（1mW）で構成された安定化光源システムです。この安定化光源は様々なオプション（SLSシリーズ）があり、SLS-110はスイッチ
光源、SLS-120はDFBレーザーで構成された安定化光源となります。

Optical Test Platforms

Optical Test Platforms
Swept-Wavelength Component Test System
The Santec’s Swept Test System has been developed to
streamline photonic testing, providing a complete solution
where high-speed analysis, high resolution and accuracy
are key. Combining one of Santec’s tunable lasers (TSL-770
o r T S L- 570 ) w i t h S a n t e c ’s o p t i c a l p o w e r m e t e r(MPM-210H), a polarization control unit (PCU-110) andcustom sof t ware, the complete Swept Test System
optimizes WDL and PDL measurement for use in both R&D
and production environments. By combining multiple TSL
units with an optical switch (OSU-110), it is possible tomeasure an ultra-wide tuning range from 1240 to 1680 nm.
Using real-time referencing, while simultaneously acquiring
output power from the tunable laser and the transmitted
optical power through the DUT, the system provides highaccuracy in WDL and PDL analysis using the Mueller Matrix
Method. Over-sampling and rescaling algorithms are used
to ma ximize te s ting thr oughpu t while ma int a ining
measurement integrity.
The Santec MPM-210H power meter mainframe can beused in conjunction with the 4- channel current meter
module, the MPM-213. The Swept Test System combinedwith the MPM-210H and MPM-213 is suitable for measuring
the per formance of fiber optics components usingtransceiver-like photodiodes (ROSA/Coherent receiver, etc.)
or optical channel monitors.

●

Achieving ultra-wide tuning range from 1240 to 1680 nmwith a full-band TSL

Applications●●●

Optical components characterization interleavers,
Fiber Bragg Gratings(FBGs), WSS, DWDM components
Photonic material characterization
Optical spectroscopy

A na l y s i sMPM-210H (Main Frame)
MPM-211 (4ch power meter)
MPM-212 (2ch power meter)
MPM-213 (4ch current meter)
MPM-215 (High dynamic range)
The MPM-210H is a new multi-port optical power meter thatis perfect for measuring the optical characteristics ofm u l t i - p o r t o p t i c a l d e v i c e s . T h e M P M - 210 H c a nsimultaneously measure up to 20 por ts (4 por ts per
MPM-211 module) very fast with high resolution. It featuresh i g h a c c u r a c y, h i g h l i n e a r i t y, a n d e x t r e m e l y l o w
polarization-dependent sensitivity with power levels from- 8 0 to +10 dBm (-70 to +5 dBm@MPM-215) ove r awavelength range of 1250-1680 nm. GPIB, TCP/IP and
RS-232C inter faces provide a convenient automatedmeasurement solution.
The MPM-210H is ideal for IL and PDL measurement ofmulti -port optical components including Dense Wavelength
Division Multiplexing (DWDM), AWG, Wavelength Selective
Switches (WSS) and more. When combined with a santec
TSL-Series laser equipped with a power monitor output, theMPM-210H allows the user to complete high precision IL
measurements while referencing in real-time. In particular,
the MPM-215 is suitable for high-speed measurement of
optical components with a dynamic range of 50 dB or more.
Seeing as the new MPM-210H multi-port optical power
meter is command compatible with the previous MPM-210
model, switching to the new MPM-210H is as simple asswapping out your devices.

IL measurement setup with the power meter MPM-210H
Power Monitor (Reference)

Computer

Trigger in●●

Wavelength Range 1250 - 1680 nm
Up to 20 ports measurement

ModulesCategory

Port number

Analog out

Power dynamic range

Dynamic range @Fixed gain

MPM-2114ch power meter2ch power meterwith analog output
High dynamic range
4ch power meter

4

None

-80 to +10 dBm

50dB (typ.)2

2 ports-80 to +10 dBm

50dB (typ.)

4

None

-70 to +5 dBm

70dB (typ.)

4

None

-70 to +10 dBm

40dB (typ.)MPM-212

OpticalPower Meter
MPM-210H

DUT

Tunable Laser
TSL-570

MPM-211/-212/-213/-215 (Modules)

Specification Table for Modules

Typical configuration

Trigger out

MPM-210H (Main frame)

MPM-215
MPM-213USB

4ch current meter

GP-IB or LAN

PCU-110IL / PDL measurement setup with the polarization controller PCU-110 and the power meter MPM-210H
Trigger out

Tunable Laser
TSL-570 / TSL-770
Full-band TSL

OpticalPower Meter
MPM-210H

DUT

Polarization
Controller
PCU-110

Computer

Trigger InPower Monitor (Reference)USB

GP-IB or LAN

The PCU-110 is PDL measurement with Mueller Matrix method.
Command control (polarization control, etc.) that can be used for
GPIB and Ether communication is possible.

Features●●●

SantecのSwept Test Systemは高速、
高精度および高分解能で光コンポーネントやモジュールのWDL/PDL特性測定が可能で、生産ラインやR&Dなど幅広い分野でご利用いただけます。
パワーモニタ出力を搭載したsantec TSLシリーズからの出力パワーモニタとDUTを透過した光パワーを同時に測定してリアルタイムにリファレンスを取得することで高精度にWDL/PDL特性を測定できます。
Swept Test Systemはsantec波長可変光源 TSLシリーズ、
パワーメータMPM-210H、
偏波コントローラPCU-110および専用ソフトウェアにて構築されます。
PDLはMuller Matrix技術により測定しています。
MPM-210HとMPM-213を組み合わせたSwept Test SystemはROSA/コヒーレントレシーバーのようなフォトディテクタを内蔵したO/Eデバイス評価にも最適です。

Wavelength range 1260-1360 / 1480-1640 nm
High SOP repeatability
Power monitor function

MPM-210Hは、マルチポート光デバイスの光特性評価のために最適な新しいマルチポート光パワーメータです。MPM-210Hは1台で最大20ポートまで同時に高速、高分解能にて測定可能です。 GPIB/Ethernetインタフェースを搭載しており、プログラムによる自動化、遠隔操作が可能です。
PCU-110は、4偏波を設定できる偏波コントローラです。Swept Test Systemに組み込むことでMuller MatrixによりPDL測定が可能になります。GPIBやEthernet通信で使用できるコマンド制御(偏波制御等)が可能になっています。

Component Testing

Component Testing

Swept Photonics Analyzer

P ol a ri z a t i on

An alysi s

SPA-100
The SPA-100 is a convenient optical frequency domainreflectometry (OFDR) add-on module to a Santec tunable
laser. When paired together, the complete system can analyze
the most compact and complex optical components yielding
results for reflectance, transmission, propogation loss anddistance to events.
The complete system employs OFDR (Optical Frequency
Domain Reflectometry) technology enabling it to analyze theback reflection and transmission characteristics of fiber optic
devices/components in the spatial domain. The systemproduces a trace like an OTDR (Optical Time Domain
Reflectometer), but with much higher resolution and precision.
The system has a sampling resolution of 5 um allowing it todiscern structures within PIC (photonic integrated circuits) andsilicon photonic (SiPh) devices with ease.
With the wide selection of tunable lasers, the OFDR system ishighly configurable to function over the spectrum ofimportance to the device being measured. The system isoffered in an O-band configuration (1260 to 1350nm range) as
well as a CL band configuration (1480 to 1640nm range). No
need to be limited to a small wavelength range.

Features
●
●

●

●
●
●

Features
●
●
●
●
●
●
●

Industry’s highest resolution reflectometer (<5 μm)
Available in O-band and CL-band configurations
Wide dynamic range (>70 dB) for WDL measurement
Proximity Sensing for Fiber to SiPh Alignment
Feedthrough port (Power monitor port) for active alignment
Easy analysis software
Configurable based on Santec TSL (TSL-570/-770/-550/-710)

Full band 1260 to 1630 nm
Simultaneous display of extinction ratio, optical power, andpolarization angle 50 dB dynamic range for PER measurement
Wide dynamic input power range(- 40 to + 10 dBm/Normal power)(- 25 to + 20 dBm/High power)
Direct PD analog output
Real-time (up to 10 Hz) measurement
Interchangeable connectors: FC, SC or LC

Applications●●●●

Alignment of polarization axis of optical devices
Connectorized PMF patch cord evaluation
PMF pigtailed laser diode evaluation
PM fiber fusion splicing

Applications
●
●
●
●
●
●

Analysis of silicon photonic waveguides
Automated SiPh wafer alignment
Optical component reflectance and analysis
Wavelength dependent component characterization
Fiber optic connector and cable assembly mating analysis
High resolution reflectance analysis and length measurement(5 μm)

Specify Multiple Refractive Indexes for a Trace

Wavelength Dependent Loss Measurement

User can specify multiple indexes of refraction within a single
trace when analyzing signals that may pass through different
mediums, i.e. from fiber to air to silicon

Measure transmission wavelength dependent loss across thefull TSL range with over 70 dB of dynamic range with no gain
switching. Single sweep and saves time.

Polarization Dependent Loss Meter
PLM-100
The PLM Polarization Dependent Loss Meter builds upon
the BRM’s stable and proven technology and takes it to thenext level. Improved source isolation, electronics, optics,
and continuous laser power referencing enhance theperformance to allow for the most repeatable and stablemeasurements.
The base model is capable of PDL, IL, and average IL
measurements with a resolution of 0.001 dB and an
unmatched PDL accuracy of up to ± (0.005 + 2% of PDL).
The backreflection option allows BR measurements from 0
to -75 dB with an accuracy of ±0.4 dB.
The PLM allows for a combination of up to 4 external inputs(1260 – 1650 nm) or internal sources (1310/1490/1550/1625nm) and a dual output option for maximum flexibility.
Custom wavelengths are available upon request.
An intuitive touchscreen display simplifies the operation.
The meter may also be controlled through remote interface(Ethernet or USB) and used with our free GMS software toautomate measurements.

Features●●●●●

Applications●●●

SPA-100は、
santec波長可変レーザ（TSLシリーズ）の便利なアドオンモジュールであり、TSLと組み合わせたシステムは、
OFDR（Optical Frequency domain Reflectometry）
技術により、
光デバイスの後方反射、および、透過特性を空間領域で分析できます。
このシステムは5μmの高サンプリング分解ののため、フォトニック集積回路
（PIC）
、
および、
シリコンフォトニクス
（SiPh）
デバイス内の構造を簡単に識別できます。

7

Ultra stable and accurate PDL, BR & IL measurements
Up to 4 internal lasers/external inputs
4 or 6 states Mueller Matrix methods
Resolution down to 0.001 dB< 1 second PDL measurements

Testing IL/PDL/BR of optical components
Incoming inspection
QA and R&D testingPEM-340は、 1260nm〜1630nmのフルバンドをカバーした偏光消光比モニタです。ノーマルパワー仕様は+10〜-40dBm、ハイパワー仕様は+20〜-20dBmの光入力に対して測定が可能です。偏光消光比・偏光角度・入力光強度を同時に測定し、結果をディジタル信号とアナログ電圧信号で得ることができます。
PLM-100は、バックリフレクションメータ(BRM-100)で実証された技術を基盤に最適化され、再現性が高く安定した測定を実現した偏波依存損失測定器です。

8

IL/RL Meters

IL/RL Meters
Optical Return Loss Meter

Insertion Loss Meter

RLM-100

ILM-100

santec's Return Loss Meter (RLM-100) incorporates 20
years of customer feedback from being industry leaders in
the cable assembly testing market. It is the next generationof optical IL/RL meters. Improvements on key hardware
components make it twice as fast as other premium testing
solutions with the same accuracy.
A new patented calibration method allows you to perform auser self-calibration and measure single-mode RL up to 85
dB with increased accuracy. The RLM 100 is ideal for
production and lab environments because of its front panel
touchscreen and easy-to-use software.
The redesigned integrating sphere detector can measure
RL loss on dense 72-channel MTP/MPO and duplex LC
connectors with a single connection. The patented SD Slide
Detector Adapters allow for the ultimate in ease-of-use.

The ILM-100 was designed to measure insertion loss on fiber
optic components quickly and accurately. The system has abuilt-in stabilized laser source for single-mode applications
or LED source for multi-mode applications. With a dual
wavelength configuration, insertion loss can be measured for
two wavelengths in less than 1 second. ILM-100 systemcomes equipped with both USB and ethernet control forease of integration into test systems.

Features
●
●
●
●
●
●
●
●

Features●●●●

Applications

Most accurate RL test in its classSelf-calibrating optical return loss meter
Wireless integrating sphere detector
Ready for production automation
Barcode control available
No computer required
Chassis modular
XNS ready

●
●
●
●
●

IL/RL testing of fiber optic cable assemblies
Single-fiber and multifiber testing
SM 1310 nm, 1490 nm, 1550 nm, 1625 nm
MM 850 nm, 1300 nm
QA and R&D testing●

Backreflection Meter

Discontinuity Test System

BRM-100

DTS-100

The BRM Backreflection Meter is an instrument developedwith extremely stable optics for precise measurement ofbackreflection, insertion loss, and power. Available with 1, 4,
12, 16, 24, 32, 48, or 72 output channels, the BRM is apractical choice for both single fiber and multifiber testing.
The BRM features up to four built-in sources at wavelengthsof 450, 650, 780, 850, 1060, 1300, 1310, 1490, 1550, 1625 or1650 nm. Custom configurations available upon request.
The BRM boasts the best backreflection accuracy in its
class (± 0.4 dB) with a measurement sensitivity to -85 dB.
Choose the integrating sphere detector option for multifiber
assemblies such as MPO or duplex LC. You can use theBRM with our free GMS software for automated testing.
The multimode BRM is internally conditioned to meet theIEC 61280-4-1 Encircled Flux standard.

The Discontinuity Test System, DTS-100, provides a method fortesting a broad variety of passive or active fiber optic
components for susceptibility to discontinuities (signal
dropouts, transient outputs or transmittance fluctuations)
during application of an external stimulus such as temperature,vibration or physical shock.
DTS-100 allows you to capture short optical power interruptions
and dropouts every 0.8 μs to 1 second. The free and easy tooperate software logs all of the events into a spreadsheet foranalysis in accordance with EIA/TIA-FOTP-32A.

Features●●●●

Stable BR measurements at low values
Up to 72 output channels
Up to 4 internal sources
Custom sources and core sizes possible

Features●●●

Applications●●●●

Testing IL/BR of fiber optic components and assemblies
Single-fiber and multifiber testing
Incoming inspection
QA and R&D testing

RLM-100は、新たに採用された特許補正方法により、ユーザーは自己補正が可能になり、最大85dBの反射減衰量を高精度で測定できるリターンロスメータです。タッチ
パネルや専用ソフトウェアなど操作性も充実しており、生産工場から研究用途まで幅広く使用いただけます。
BRM-100は、後方反射、挿入損失を正確に測定するために、非常に安定した光学系で構成された反射減衰量測定器です。最大72の出力チャネルで利用可能で、シングル
ファイバーとマルチファイバーの両方の評価に対応しています。

9●

Applications

Compact benchtop instrument for all-in-one operation
USB and Ethernet interface
Test software OPL-CLX available for logging measurements
Fully automated single and dual wavelength insertion
loss measurements
Multi-mode instruments available with define launch conditions(EF, AS100, 70/70, etc.)
Various detector options

●
●
●
●

Single-mode and multimode
Input power range of -40 dBm to -2 dBm
Selectable dropout detection range of 0.8 μs to 1 s
Selectable loss threshold of 0.5 dB to 3 dB
Bundled application that records data directly into a spreadsheet
USB interface for seamless remote control
Support of most common connector options (FC, ST,SC, etc...)●●●Cable assembly testing
Optical alignment
Signal monitoring

Applications●●●

Optical component qualification and validation
Stress and reliability testing
Compliance testing such as EIA/TIA-FOTP-32A or similar

ILM-100 は、光コンポーネントの挿入損失を迅速かつ正確に測定する挿入損失測定器です。
DTS-100 は、光ファイバーや光部品に温度や振動などの外部刺激が加わった際の不連続性 (信号ドロップアウト、過渡出力、または、透過率の変動) を評価可能な不連続性評価システムです。0.8 μs〜1 秒ごとの短い光パワーの中断やドロップアウトを確認できます。

10

Optical Power Meters

Optical Switches

Multichannel Optical Power Meter

Optical Switch
OSX-100

PM fiber available

The OSX Optical Switch, OSX-100, is a customizablebenchtop/rackmount instrument ideal for high-volume
manufacturing production testing.Available in single-mode or
multimode, channel counts range from 1x2 up to 1x400.
Extra common inputs are possible by adding a 1xN or in 2B or
2C configurations. A variety of other path routing configurations
are available upon request.
The switch can be controlled through USB or Ethernet. The
front panel bulkheads can be fitted with FC, SC, or LC
connector types.

Features
●
●
●
●
●
●

Up to 400 outputs
Ultra low IL < 0.5 dB± 0.005 dB repeatability
Crosstalk < -80 dB
Customization available
Available in SM, MM, PM and custom core sizes

Applications●●●●

High-volume production testingMultiple device testing
Test automation
Pair with RLM for multifiber testingOPM-150
Santec's OPM-150 Multichannel Optical Power Meter is acost-effective solution for manufacturers or labs requiringhigh channel counts. Available with up to 24 individual
detectors, the OPM-150 includes simultaneous power
measurements on all channels and built-in USB and
Ethernet communication.
The OPM-150 is capable of faster than 30 ms sampling timeover the USB interface allowing for fast feedback in active
optical alignment applications for fiber coupling and silicon
photonics. It's simultaneous optical power measurement
capability enables users to instantly spot a port failure,
making the OPM-150 an ideal instrument for demanding
production and lab applications. Software is available foreasy automated testing and data export.

Features
●

●
●
●
●
●

Mix and match different detector types (Si or InGaAs) andsizes (1, 3, 5 or 10 mm)
Simultaneous readings from up to 24 detectors< 30 ms sampling time
Colour touch screen display
USB or Ethernet
Replaces OP760

Applications
●
●
●
●
●
●
●

Optical alignment
Silicon photonics
Optical signal monitoring
Environmental testing
Transceiver testing
Lab and R&D
Parallel laser burn-in testing

OEM Optical Switch

High Performance Optical Power Meter

OSX-100E

OPM-200
PM fiber available

The OSX-100E is an off-the-shelf fiber optic switch component
that can be easily integrated into virtually any fiber optic routing
or switching application. It is readily available in common fiber
types (standard single-mode, 50 um or 62.5 um multimode); but
can also be ordered with specialty fiber types (step index, large
core multimode, small core single-mode).
Switches can be configured for 1x2 up to 1x80 channels andare also available in 2B or 2C configurations (see configuration
options). For more complex signal routing and switching, thelow loss (< 0.5 dB) allows for concatenating switches to enablehigher channel counts. For ease of integration, each switch
comes with a pre-programmed driver board and uses standard
I2C communication protocol for control. The driver board thatcan be mounted separately or with the optical switch, giving
more flexibility.

Features
●
●
●
●
●
●

Up to 80 outputs
Ultra low IL < 0.5 dB± 0.005 dB repeatability
Crosstalk < -80 dB
Customization available
I2C communication

Features●

Applications●●●●

Fiber optic signal routing/conditioning
Automation
Instrumentation
Fiber optic sensing

OSX-100は、信頼性が高くカスタマイズ可能なベンチトップ型の光スイッチで、研究用途から生産工場用途まで幅広くご利用いただけます。シングルモードまたはマルチモードで利用でき、チャネル数は 1x2 から 1x400までの幅広いラインアップを揃えています。
OSX-100E は、あらゆる光ファイバルーティング、または、スイッチングアプリケーションに簡単に統合できるOEM用光スイッチです。一般的なファイバタイプ (シングルモード
ファイバ/50 μm、または、マルチモードファイバ/62.5 μm) に加え、特殊ファイバタイプ (ステップインデックス、ラージコア/マルチモード、スモールコア/シングルモード) にも対応しています。

11

The OPM-200 High Performance Optical Power Meter is thelatest generation of Santec power meters. The standard 2
mm InGaAs detector can measure power down to -80 dBm
over a wide wavelength range from 840 to 1700 nm.
Superior accuracy and 50 μs averaging time make theOPM-200 a reliable power meter for almost any application.
The detector is free-space and compatible with ourpatented SD Slide Detector Adapters. Together with theanalog output it is ideal for automated optical alignment.
Additionally, choose from 1 to 4 detectors and remote-head
options for the ultimate, flexible power meter solution.

●
●
●
●
●
●
●

Free-space 2 mm InGaAs detector
FC, SC, LC, MPO and more detector adapters are available
840 to 1700 nm wavelength range+30 to -80 dBm power range
50 μs averaging time
Touch screen display
USB or Ethernet
30 Day Money back guarantee

Applications
●
●
●
●
●

Optical alignment
Transceiver testing
Laser and amplifier characterization
Transient testing
Lab and R&D

OPM-150は、最大24チャンネルを同時に測定できるコストパフォーマンスの高いマルチチャンネル光パワー メータです。USB インターフェイスにより 30 ms の高速サンプリン
グが可能で、ファイバー結合やシリコンフォトニクスのアクティブな光アライメントなどのアプリケーションに最適です。
OPM-200は、広い波長範囲と高いダイナミックレンジを有した光パワーメータです。特許取得済みの SD Slide Detector アダプターと互換性があるだけでなく、フリースペースにも対応しており、幅広い用途で利用していただけます。

12

Polarity Testers

Optical Attenuator
Optical Variable Attenuator

Polarity Test Meter

OVA-100

PTM-100

Santec's Optical Variable Attenuator, OVA-100, enablesprecise optical power control featuring high accuracy andsuperior repeatability. It can be used as a benchtop or
rackmount instrument. The OVA-100 is ideal for lab andproduction applications including power level adjustments
in automated test systems, Bit Error Rate Testing (BERT) of
transceivers, and channel equalization in WDM systems.
The OVA-100 attenuator offers ± 0.01dB repeatability and a
dynamic range of up to 100 dB (depending on the model
configuration). OVA-100 can be controlled via the front panel
touch screen, or remotely via USB and Ethernet. OVA series
attenuators operate from 1200-1700 nm for single-mode
and 750-1700 nm for multi-mode. A power monitoring PD
can be added to read the power output of the instrument. It
can be used to auto adjust the attenuation for a constantcustomized power level.

The PTM Polarity Test Meter is an easy-to-use instrument
specifically designed to check the polarity of multifiber
cable assemblies. The PTM is equipped with a touchscreen
inte r f a c e to a llow for one touch te s ting of t y pic a l
pre-programmed types A, B, C, and unlimited configurable
custom profiles. The Visual Fault Locator ( VFL) mode
enables easy manual identification of any possible damaged
or crossed fibers.
The PTM can be built with up to 288 channels in one
chassis and expanded to 432, if needed. The PTM can
operate on a stand-alone Santec platform which does not
require a PC to operate. There is still the possibility of
remote operation through USB or Ethernet. You can use thefree Cable Assembly software to save your polarity results
along with your IL/RL results. The included PTS software
package also allows you to test just polarity with results
exported to Microsoft Excel.
The PTM has been developed with customers in mind bylooking at the growing demand to reduce testing time andto increase throughput with a user-friendly experience.

Features
●
●
●
●
●

●

Precise optical power± 0.1 dB accuracy± 0.01 dB repeatability
100 dB dynamic range
Available in single-mode 9 μm, Panda PM, HI 1060,
Multimode 50 μm and 62.5 μm
Custom fiber types upon request

Applications●●●

Manufacturing production testing
Bit Error Rate Testing (BERT)
Channel equalization

Features●●●●

Fast Pass/Fail results
Touchscreen display and operation
Up to 288 channels in one box
Test results stored with XNS IL/RL results or to Excel

Optical Backreflector

Remote-Head Polarity Detector

OVB-100

RD-P-100

The OVB Optical Variable Backreflector enables precise
backreflection control and superior repeatability. It can beused as a benchtop or rackmount instrument. The OVB is
ideal for lab and production applications includingtransceiver testing, QA testing, laser production andautomated test systems. The OVB variable backreflectorof fers ± 0.02dB repeatability and a range of 60dB insingle-mode and 35 dB in multi-mode. The OVB can becontrolled via the front panel touch screen, or remotely via
USB and Ethernet. The OVB series variable backreflectors
operate from 1200-1700 for single-mode and 750-1700 for
multi-mode. A 50/50 coupler can be added to bypass thevariable backreflector for input signal monitoring.

Create the ultimate multifiber cable assembly test station byadding the RD-P Remote-head Polarity Detector to yourRLM Return Loss Meter and OSX Optical Switch. In a single
connection, you can now test IL, RL and polarit y ofmultifiber assemblies such as MPO or duplex LC.
Use with our free Cable Assembly software and have all
your IL, RL and polarity results saved together in the XNS
database.

Features●●●●±0.2 dB relative accuracy±0.02 repeatability
60 dB range
Available in single-mode or multi-mode

●
●
●
●

MTP/MPO cable assembly mapping
Multifiber assemblies
Complex network mapping
QC inspection

Features●●●

Applications●●●

Transceiver testing
QA testing
Laser production

OVA-100は、高い精度と再現性を備え正確なパワー制御が可能なベンチトップ型の光可変減衰器です。シングルモードでは 1200 〜 1700nm、マルチモードでは 750 〜
1700nm の波長範囲に対応しています。
OVB-100は優れた後方反射制御と再現性（±0.02dB ）を有する光可変後方反射測定器です。その測定範囲は、シングルモードで 60dB、マルチモードで 35dBに対応しており、研究用途から生産用途まで幅広い用途にご利用いただけます。

13

Applications●●

Test polarity then IL/RL without disconnecting your DUT
Fast Pass/Fail results
Can measure most multifiber connectors
Reduce reference complexity
Test results stored with XNS IL/RL results

PTM-100は、マルチファイバーケーブルアセンブリの極性をチェックするために使いやすさを追求し特別に設計された極性テスタです。タッチスクリーンインターフェイスによりPC を必要としないスタンドアロンとしても動作が可能です。
RD-P-100 は、デバイスを切断することなく、12本のファイバーをわずか4秒で測定が可能なリモートヘッドタイプの極性テスタです。RD-P-100をRLM (リターンロスメータ) とOSX(光スイッチ)に接続することで、究極のマルチファイバーケーブルアセンブリ評価用ステーションとなります。

14

Launch Conditioning

Tunable Filters
Launch Condition Meter
LCM-100
The LCM Launch Condition Meter is a convenient benchtop
instrument for analyzing characteristics and geometry of
optical fibers. Use Santec's software to scan and easily plot
both the near-field and far-field patterns. Compare to
standard launch templates such as the IEC 61280-4-1
Encircled Flux standard. The LCM can be equipped with
two internal LEDs with 850 nm and 1300 nm wavelengths.
Internal 105/125µm, 0.22NA fiber allows for overfill testing inmost applications. The optical interface of the instrument
accepts all standard 2.5mm ferrules. Other connector sizes
are available upon request.

Features●●●●●

Applications

Near-fieldFar-field measurements
Auto-alignment and focusing
Mode Field Diameter measurements
Software to compare to common launch condition
templates such as Encircled Flux●●●

Fiber analysis
Insertion loss source compliance testing
R&D

O-band is available upon request.

Mode Fill Conditioner
MFC-1XXThe OTF-350 is a versatile manually optical tunable filter
that allows simultaneous and independent tuning ofcenter wavelength and bandwidth over C & L bands. Its
unique precision free-space optics and accurate tuning
mechanism bring forth excellent optical control and a flat
top passband shape of filtered signal.

Features●●●●

Wavelength tuning range: 80 nm
Wavelength range: 1530-1610 nm (C&L Band)O-band is available upon request.
Bandwidth: 0.2-10.0 nm (Normal type)
Bandwidth: 0.1-15.0 nm (Wide type)

The Santec Mode Fill Conditioner, MFC, is an accessory to
any compatible Santec instrument used for measuring
insertion loss and is essential for guaranteeing that theproper launch conditions are met when testing insertion loss
on multi-mode fiber optic assemblies. The MFC can be
configured for almost any industry standard specified launch
condition such as telcom standard encircled flux or military
standard AS100, AS62. Each MFC can be configured withvarious standard connector types (FC, LC, SC, ST) and inboth APC and PC polish types. The system is available in asingle launch conditioning cable (MFC-100) or a rack mount
multi-channel system (MFC-150).

Features●●●●

OTF-980は、波長と透過帯域幅をそれぞれ独立して変えることができるベンチトップ型の可変光フィルタです。回折格子をベースとしたユニークな高精度可変メカニズムにより、フラットトップな透過フィルタスペクトルと優れた光学特性を実現しております。波長と透過帯域幅の設定はタッチスクリーンもしくはGPIB/Ethernet/USBインタフェー
スより行うことができ、プログラムによる自動化、遠隔操作が可能です。
OTF-350は、波長と透過帯域幅をそれぞれ独立して変えることができる手動型可変光フィルタです。

15

Customizable Launch condition
Many fiber sizes/types
Common connector types available
Available in compact single fiber patch cord

Applications●

Standardized insertion loss testing

LCM-100は、光ファイバーの特性と形状を分析するための励振状態評価機です。 専用ソフトウェアにて、簡単にニアフィールドとファーフィールドの両方のパターンをスキャン
してプロットします。 IEC 61280-4-1 Encircled Flux 規格などの標準励振テンプレートと比較できます。
MFC-1XXはマルチモードファイバの挿入損失測定時に適切な励振状態にするために必要な光ファイバモード整形器です。各種通信規格、Encircled Flux 規格、MIL規格(AS62, AS100) に準拠します。

16

Our LCOS-SLM (liquid crystal on silicon based spatial light modulator) is an advanced spatial light modulator for full high resolution
applications. WUXGA (1920 x 1200) resolution, 10-bit (1024 gray levels), and excellent phase stability (~0.003π rad.) make ultra-fine light
modulation possible for various applications.

High-Speed model

SLM-210
FeaturesResponse time typ. 6 ms, (Type A)
WUXGA (1920 x 1200)
Memory function
Triggers-input & output
Robust long FPC

Laser processing (500W durable model)

SLM-310-G
Features
Wavelength Range 532 ± 50 nm
High-power model holds the world record
for high-power durability up to CW 500 Watts.
LCOS head with water-cooled heat sink

Applications
Holography
Wavefront correction
Beam steering
Biosensing
Quantum computing
Pulse/Beam shaping
Programmable phase pattern

Applications
Laser processing
3D printing
IC trimming

Current model / 249 ms

High speed model / 6 ms

249 ms< 6 ms

140

800

120

700
600

80

Optical power [a.u.]

Optical power [a.u.]

100
60
40
20
0

300
200
100
0

0

200

400

600

800

1000

Time [msec]

Current model

0

5

10

15

20

Time [msec]

High speed modelHigh speed model

SLM-310-G は当社第 3 世代となる液晶 LCOS 技術を用い、金属加工に有効なグリーンレーザーにおいて光出力パワー 500W の耐光性を実現しました。一般的な金属では、グリーン波長の吸収は近赤外波長の 1μm と比べて 10 倍以上大きく、レーザー加工で有効な波長帯と知られています。また、従来製品同様 10bit（1024 階調）の高分解能特性や高い位相安定度特性もそのまま継承しております。

17

400

-100

-20

Current model

500

SLM-210 は、当社第 2 世代の液晶 LCOS（Liquid Crystal on Silicon）技術を用いた高性能製品として、応答速度を大幅に 向上させた最新のハイスピードモデルです。
従来品と比較して 20 倍以上の高速化となる、応答速度 10 ms 以下を実現。LCOS の応答速度向上に伴い当社特有の優れた 高階調度と相まって、光波面補正、レーザー
加工用光ビーム 整形、バイオセンシング、量子コンピューティングなどの光応 用分野での性能向上に貢献します。

18

Comparison Tables for Spatial Light Modulator
Standard model

SLM-200/20
Wavelength range 450〜1600 nm
WUXGA (1920 x 1200) resolution
10-bit (1024 gray levels)
Excellent phase stability (~0.001π rad.)UV Hardened model

SLM-250
Wavelength range 365〜550 nm
WUXGA (1920 x 1200) resolution
42 times higher UV tolerancecompared to the visible to InfraRed (IR) wavelength SLM-200
10-bit (1024 gray levels)

Item

SLM-210SLM-310-G

Wavelength range
Response time 1)

Type A

SLM-300/30

365-550

450-1600

nm

Typ. 200

Typ. 50

Typ. 200

ms

Typ. >90

Typ. >92

Typ. >70

Typ. >90%

Phase stability

Typ. < 0.003π

Typ. < 0.002π

Typ. < 0.003π

Typ. < 0.003π

Typ. < 0.001πrad.

Max. 500

Typ. 10Max. 200

Typ. 40 MW/cm2

Typ. 10

W/cm2

8

-

1-2

-

-

L/min.

Water inlet and outlet

Pipe ﬁttings-

Pipe ﬁttings

-

-

-

Dimensions LCOS unit

60×85.2×21.8

45 x 45 x 25.7

60 x 60 x 31.7

45 x 45 x 25.7

45 x 45 x 25.7mm

Dimensions SLM body

122 x 122 x 38.3

117.6 x 117.6 x 33.7

122 x 122 x 38.3

122 x 122 x 38.3

117.6 x 117.6 x 33.7mm

Dimensions Drive board--

70 x 70 x 1.6-

70 x 70 x 1.6mm

Dimensions Option board--

95 x 101 x 1.6-

95 x 101 x 1.6

mm

Optical power handling 2)
Water ﬂow

(H)15.36 x (V)9.60

Panel sizemm

Panel resolution 3)(H)1920 x (V)1200

pixel

Pixel size / pitch

7.8 / 8.0μm

Aperture ratio

95%

Gray level

10 (1024 levels)

bit

Frame rate

60 or 120Hz

LCOS drive frequency
Phase depthInterface 4)

1200

Hz

Min. 2πrad.
-

DVI / USB 3.0

Operating temperature

℃

15-35

Storage temperature

℃

0-40

-

GUI software and SDK for Windows: C#, Python, Matlab, Labview

Wavelength Range
300

400

500

600

700

1000

1100

1200

1300

1400

-02 750 - 850-03 1000 - 1100

(nm)

1600

1700

-04 1500 - 1600

-14 450 - 550

SLM-201500

400-700-12

SLM-200

900

NIR-01 450 - 550

SLM-210800

VIS-14 1500 - 1600-21

450 - 1600

365 - 550
-02 750 - 850-01 485 - 580

SLM-310-G

We can custom-build special SLM to suit your needs.
With heater, temperature sensor, etc.

532, 800, 1064

450-1100

Tr:17 / Tf:53 Tr:48 / Tf:200

Typ. >92

SLM-250

Custom & Special Orders450-1600

450-700
Tr:6 / Tf:18

Panel reﬂectivity

SLM-300, SLM-30

Other models

SLM-200
SLM-20 5)

Typ. 200

UV

Wavelength Range 532, 800, 1064 nm
20 times higher power durability to SLM-200
LCOS head with water-cooled heat sink
10-bit (1024 gray levels)

SLM-250532 ± 50

Control software

High-Power modelType C

SLM-300
SLM-30 5)

Units

Type B-03 1020-1110

485 - 580

< AR coating option for SLM-210, 200, 20>
Item
AR coating range 6)
AR coating reﬂectance 7)

-00

-01

-02

-03

-04

-12

no coating

450-550

750-850

1000-1100

1500-1600

41) Response time is a typical value and is not aﬀected by frame rate.

Tr: Rise time between 10% and 90% levels in a phase change of 0 to 2π
rad. at 25°
C. Tf: Fall time between 90% and 10% levels in a phasechange of 0 to 2π rad. at 25°
C.

2) The value is not guaranteed.

SLM-210, 200, 20: 1550 nm CW, 2.0 mm beam diameter
SLM-300, 30: CW @1064 nm

SLM-250: Typ. 40 MW/cm2 : Peak power @355 nm, Pulse laser.

< 0.5

-21

Units

400-700

450-550 /
1500-1600-14450-1600

nm

< 1.5

< 0.6

< 2.5

%

3) Speciﬁcation on the defect pixels are no object.
4) DVI: 10-bit using RGB 8-bit, 3 colors

5) The SLM series consists of an LCOS unit and SLM body, but only

embedded module SLM-30,20 consists of LCOS unit, Drive board andOption board.

6) We support custom AR coating request. Please contact us for detail.
7) Angle of incidence = 0 degree

SLM-250: Max. 10 mW/cm2 : @365 nm, 24H/day continuous operation.

SLM-200 は､ビーム整形､波面補正､光マニピュレーションなど､さまざまな科学･工業用途に適した空間光変調器です。SLM-250 は､液晶及び光学薄膜構造の最適化により､優れた紫外線耐性を有しています。SLM-300 は､水冷ヒートシンクを備えた LCOS ヘッドを採用し､200W( ワット ) クラスの高出力レーザーにご利用いただけます。
SLM-20/SLM-30 は､産業分野における応用展開が出来るよう､小型･低価格化した機器組み込み用の空間光変調器です。SLM シリーズはいずれもパターンジェネレータ
ソフトウェアが付属しており､コンピュータで生成したホログラムの計算が可能です｡また､各種モデルはお客様のご要望に合わせた柔軟なカスタマイズに対応しています。

19

20

Programmable Optical Filters
Programmable Arbitrary Spectral Shaping

Multichannel Optical Power Monitor

WSS-2000
The WSS-2000 is a LCOS based Programmable Optical Filter. Santec's proprietary LCOS (Liquid Crystal on Silicon) technology inside the
WSS-2000 offers the ultimate in flexibility, enabling the user to program a limitless range of optical filtering, attenuation and switching
schemes. The WSS-2000 is an enabling technology for investigating next generation optical networks by utilizing the filter as an opticalequalizer of high-speed signals, as an adaptive channel filter for advanced optical transmission systems (DWDM, OFDM), as a WSS emulator
or as a flexible test and measurement system.

Features
Programmable arbitrary spectral generation and spectral shaping
Fine frequency and bandwidth control using LCOS. Setting
resolution 0.78GHz (typ.)Excellent optical filtering with steeper edge 400dB/nm (typ.)Switching function optionally 1, 2, or 4 outputs

The MPA-100 is a highly integrated rack model of pigtailed
photo-detector array. Multi-channels optical power from MPO
or MTP connectors are detected by photo-detector array.
Photo-detector of Si type for 850 nm range and InGaAs typefor 1260~1610 nm are available. The detected photo-current isconverted to voltage output using a 16-bit DAQ interface. The
dimension is ( W )4 82. 6x(D)430 x(H)4 4 mm. T he highly
integrated MPA-100 is available up to 80 channels. The
MPA-100 is useful for many applications, such as testing andmonitoring of transceivers for data communications in data
centers, broadband/CATV, and telecommunications. Tap-PD
model (fibers input&output, a partial power detection) is alsoavailable upon customer’s request.

Built-inPrivate Brand
PD

Standard model

FeaturesMulti-channels photo-detector array up to 80 channels
Power monitoring and testing at 850 nm SR4, SR10
Wide-operating wavelength range: 1260 ~ 1610 nm
Wide-optical input range: -50 ~ -10 dBm
Excellent linearity: +/- 0.6dB
MPO adapter / High-density MPO, MTP connectors
16-bit DAQ voltage output by a USB interface
1U Size (44mm) slim design
Custom model is available upon request

Applications
Optical amplifier testing and evaluation with opticalequalizer for high-speed optical signal
100Gb/s, 400Gb/s high speed transmission test
Adjustable and adaptive DWDM,OFDM channel filtering
Flexible test and measurement
Next generation bundle wavelength OXC
Pulse shaping
Optical comb generation
Wavelength Selective Switch (WSS) emulator

Measurement Data

Built-inPrivate Brand
Tap PD

Normalized loss [dB]037.5 GHz
50GHz
100GHz200GHz

-10

-20

-30

-50
-200

-150

-100

-50

0

50

100

150

Applications

Custom model

Optical power testing of transceivers for IEEE 802.3ba based SR4 ,SR10, LR4, ER4
Optical power monitoring for data communications, broadband / CATV networks,
and telecommunication networks
Optical power monitoring in data center equipment
Optical power monitoring of Interconnections of active optical cables

200

Relative frequency from ITU-grid center [GHz]

Normalized loss [dB]

-4

BW3 Setting
42.5 GHz
42.0
41.5
41.0
40.5
40.0
39.5
39.0

-5

-6

-7

-8
194,166

194,168

194,170

194,172

194,174

Frequency [GHz]

Bandwidth and frequency tuning
Current modelHigh speed model

WSS-2000 は任意の光スペクトルを生成するためにプログラム可能な LCOS をベースとした光フィルタです。当社 LCOS 技術は 10bit（1024 階調）の高分解能と低いフリッカーノイズが特徴であり柔軟性に富んだ帯域可変、減衰制御、スイッチング動作を実現しております。WSS-2000 は次世代光ネットワークの評価及び検証に適しており、高速信号用の光イコライザ、DWDM（光波長多重通信）や OFDM（直交周波数分割多重方式）チャネル選択フィルタ、WSS エミュレータとして、各種測定にご利用いただけます。

21

MPA-100 はピグテイル型フォトディテクタ・アレイのラック・マウント型製品です。MPO・MTP 多芯コネクタより入力される複数の光パワー信号をモニタします。
波長 850 nm 帯を検出するシリコンタイプと波長 1260 〜 1610 nm 帯を検出する InGaAs タイプの光検出器よりお選び頂けます。光電流は内蔵する 16 ビットのマルチ
信号 DAQ インターフェースを介して、USB 端子より電圧出力されます。

22

Optical Monitors /Variable Attenuators

Optical Filters
MEMS Tunable Filter
MTF

Santec's UTP-1-LL and UTP-1 are the world’s smallest single,
high directivity, uni-directional tap monitors.
Compared to the current Santec uni-directional MTP-1, the new
UTP-1 is an 80% reduction in size, allowing the customer toreduce their module’s footprint while increasing design
capability.
Santec's tap monitor series is excellent for high-densit yintegration in next generation small form factor packages
matching CFP2 , QSFP-DD and OSFP

Santec's MTF is the world's smallest tunable filter based on electrostatic MEMS technology. Especially our MEMS have excellent shockand vibration resistance.

MTF-1 (1 Gen.)st

Features
UTP-1

Applications
ASE Noise Reduction in EDF Amplifiers
400G+tranceivers in CFP2 , QSFP-DD and OSFPHighly reliable amplifiers
Optical channel monitors

World’s smallest size 7 x 5 x 3.5 mm
Minimum fiber bending radius of 5 mm
Robust mechanical design in shock and vibration
Low drive voltage 12 V

MTF Spectrum
1523.2

World's smallest size
Excellent characteristics
Telcordia GR-468 qualified
Pin bending is available
Minimum fiber bending radius of 5 mm is available
80/125μm dual-core fiber support

1535.2

1543.71568.81552.10-5Insertion Loss [dB]

-10
-15
-20
-25
-30
-35

Power and channel monitoring
Telecommunication systems
Switch and protection
DWDM systems
Amplifier
CFP2,QSFP-DD and OSFP

MTF-1

UTP-24

-40
1510

1530

1540

1550

1560

1570

1580

Wavelength [nm]MTF-FT

MTF-VB(2nd Gen.)

Features

World's first commercialization
Flat-top frequency response by excellent gaussian order
Ultra-Compact Size 7.5 x 5.5 x3.5 mm
Low Voltage Drive 17 V

MEMS Attenuator

1520(3rd Gen.)

Features・World's first commercialization
・Continuous λ& BW independent control
・Flat-top frequency response by excellent gaussian order
・Ultra-Compact Size 9 x 5.5 x 3.5 mm

COVA-1
Santec's COVA-1 is the world's smallest optical variable attenuator
based on electrostatic MEMS technology. Compared to the current
Santec MOVA-1, the COVA-1 has achieved an 82% reduction involume ratio. Santec's variable attenuator series is excellent for highdensity integration in next generation small form factor packages.

COVA-1
MTF-VB

MTF-FTUTP シリーズは､世界最小サイズの高いダイレクティビティを持つユニディレクショナル光モニタ部品です。
体積比で従来モデルの MTP1 から 80% の低減を実現し、CFPx に代表される SFP (Small Form-factor Pluggable) 標準規格に適合する製品サイズを実現しております。極めてコンパクトなパッケージ設計が特長で､分岐カプラを設ける必要がありません。C バンドから L バンドまでの広い波長範囲で低損失､高感度な優れた特性を持ちます。

10

20
25
30
35
-1000

5
10

1

15

-500

0

500

Normalized Frequency [GHz]1000

Normalized Loss [dB]

Normalized Loss [dB]5World's smallest size
Bright / Dark types available
Low power consumption
Telcordia GR-1221 qualified
Shock and vibration tolerance design
Minimum fiber bending radius of 5 mm is available
Shutter function

0

0

0

2

Insertion Loss [dB]

Features

BOX-like profile

3
4
5
-200

-100

0

100

200

Normalized Frequency [GHz]

15
20
25
30
35
40
45
190

191

192

193

194
195
196
Frequency [GHz]

197

198

MTF は、MEMS(Micro Electro Mechanical Systems) 技術に基づく世界最小サイズの波長可変フィルタです。高速大容量光伝送において光増幅後の雑音光除去に用いられる重要な光部品であり、光トランシーバの小型集積化に貢献します。従来のガウシアンフィルタに加え、800Gb/s 越の超高速信号用に透過帯域が平坦、かつ急峻な雑音除去性能を特徴とするフラットトップフィルタ (MTF-FT)、及び 400Gb/s から 1.2Tb/s の Flex rate に対応した帯域幅・波長可変フィルタ (MTF-VB) がございます。

COVA シリーズは、PDL や波長依存性が小さく波長多重通信で必要とされる各種仕様について優れた特性を持つ超小型可変アッテネータです。MEMS(Micro Electro
Mechanical Systems) 技術に基づく内部ミラー設計が採用されており、電圧変動の繰り返しや、大きな環境温度変化、機械的な衝撃に対して耐久性の高いデバイスです。

23

24

We want to improve the eyesight of as many people as possible. That's why we've built our company onsuperior engineering. Based in Santa Clara, CA, we leveraged our extensive experience in cutting-edge
optical science. By focusing on creating advanced optical imaging techniques, we help physicians
improve the speed and accuracy in diagnosis.

25

26

IVS-1000(1060 nm), -2000(1310 nm)

OCT System Line up
Model name and Option

Center
Axial
wavelength(nm) resolution(μm)

Standard (ST)Max. Imaging
depth (mm)Image acquisition Scanningspeed (lines/sec) Laser Type

<18

>4

20,000<9>420,000Long Imaging depth (LC)

<18

>12

50,000High Speed (HS)

<18

>10

100,000<16

20 to 150

10,000 to 400,000

High Resolution (HR)IVS-2000

Polygonscanner

1310
MEMS

IVS-1000

Standard (VCSEL)

1060

TunableVCSEL

IVS-2000は、研究用途、製品化の為の基礎検討などさまざまな分野・応用でご利用いただけるように、光源やプローブをご希望にあわせてお選びいただけるカスタマイズ対応が可能なSS-OCTシステムです。オプションでLabViewを用いたコード開示も出来ますので、お客様ご自信でのカスタマイズも可能です。LabView, C#, C++用のSDKもオプションで提供しております。

27

28

OCT Scanning Lasers

OCT Scanning Lasers●●●

OEM source for OCT systems
Spectroscopy and Optic Sensing
FMCW LiDAR

65
60
55
50
45
40
35
30
25
20
15
10
5
0

1120
1100
1080
1060
1040

Wavelength(nm)Applications

Output Power (mW)
Trigger Voltage (V)HSL-1 Optical Characteristics @200 kHz scan rate

1020
0

1

2

3

4

5

Time (us)

6

7

8

9

1000

Type

Tunable VCSEL
Flexible scan rate, Ultra long coherence length, No coherence
revival.

MEMS
High speed, Long coherence length

Polygon scanner
Simple rescaling, High accuracy

1060 nm
Ophthalmology OCT

1310 nmConventional OCT

1550 nmSpectroscopic OCT3000 nmSpectroscopic OCT

fs: 4-400 kHz
⊿ λ:
⊿
λ: 75
80 nm
nm
Lc: >1 m-

fs: 10-400 kHz⊿ λ: 80 nm⊿
Lc: >1 mcoming soon

fs: 10-200 kHz⊿λ: 80 nm⊿λ:Linewidth: <10 MHz

fs: 100 kHz⊿λ: 100 nm
Lc: 15 mm

fs: 50, 100 kHz⊿λ: 100-130 nm
Lc: 10-18 mm--

fs: 20-50 kHz⊿λ: 100-170 nm
Lc: 2-5 mm--*Lc：Coherence length

Santecの光源技術を駆使した高速波長掃引レーザーHSLシリーズ。狭線幅のレーザーが連続的に高速スキャンします。ターンキーひとつで起動し、トリガーも内蔵しているため、シンプルに測定系を構築する事が可能です。RIN（Relative Intensity Noise）が小さいため、OCTで利用した場合に高いS/Nを得ることが出来ます。OCTだけではな
く、高速な分光分析用の光源としてもご利用いただけます。HSL-1は面発光レーザーの技術を応用し、シングルモード発信で高速に波長をスキャン。帯域やスキャンレート
を可変できます。HSL-20はMEMSフィルタを用いた外部共振器型レーザーで、OEM光源として広く使われております。

29

HSL-2100はポリゴンスキャナを用いた外部共振器型レーザーで、高繰り返し性、直線性を有しており、容易に波長校正が可能です。特に高精度の計測などで使われております。HSL-200はHSL-2100のOEM版で、コンパクトな設計、外部からのシリアル通信による制御が可能です。

30

DC to 80MHz(Suitable for <30kHz A-Line rate)
DC to 200MHz(Suitable for >30kHz A-Line rate)
DC to 400MHz(Suitable for >100kHz A-Line rate)

Parameter

Unit

Center Wavelength

Specification
HSL-1

HSL-10

HSL-20-50

HSL-20-100

nm1045 - 1075

1040 - 1070

1280 - 1340

1280 - 1340

Axial Resolution (in air)

μm

<15

<18

<18

<18

Imaging Depth Range (in air)mm

>70 (@10kHZ)

>10

>20

>10

A-line Rate

kHz

10-400 +/- 0.1

100 +/- 0.1

50 +/- 0.1

100 +/- 0.1

Peak Output Power

mW

≥40

Optical Fiber Type

-

SMF

OS

-

Windows 10 (64 bit)

System Bus-

PCI Express 2.0 (Gen2) 5.0 GT/s x8*Please note, these specifications are subject to change. Please contact us for the details.

3D OCT Viewer Software

OCTEngineは、高速波長掃引レーザーHSLシリーズ、バランスドフォトディテクタBPD-200、及びリアルタイムサンプリングDAQカードHAD-5200を使用したSS-OCTを実現するための核となるパーツです。

31

32

3D Optical Profiling

3D Optical ProfilingHigh Sensitive and High Speed Game Changing
3D Optical Profiler for Production Development Support

OPS-1000 Software
Our data acquisition interface is included with the
OPS-1000 for convenience. The software provides
3D Point Cloud Data and 3D Light Intensity Data forpart analysis. The data is output as a PCD (Point
Cloud Data) and the software is compatible with
Windows OS.

The Santec Optical Profiling System uses the
combination of a high-speed tunable laser and aninterferometric detection technique to bring record
sensitivity to a single beam, non-contact opticalprofiling. The Santec OPS-1000 is designed todeliver high speed, high accuracy and non-contact
measurements regardless of shape, material, coloror surface shape of the target object.

Measurement Examples

Top 3 Features of OPS-1000
1. High Sensitivity

Minimum light detection sensitivity of 100dB. Dynamic Range > 70dB・Diameter of holes
Based on the measured point cloud data, you can find theedges of the hole in the part and measure the diameterand depth.
3D Height Map Data

Drill Bit

2. High Speed Measurement

Automate parts inspection with high speeds of up to 400,000 points / second

・Distance between holes
You can measure the distance between multiple holes.

Gear Box

3D Height Map Data

3. Single Beam Geometry

Minimized structural interference within complex parts

・Comparison with CAD Data
By reading the measured point cloud data and CAD data,
the difference value can be displayed in a colored map.

Engine Cylinder Head

3D Height Map Data CAD Comparison Data

OPS-1000は高速波長可変レーザの技術を干渉計測に応用した、世界最高感度の光学式三次元計測器です。本製品はヘテロダイン検波方式により、約100dBという、三次元計測器では世界最高レベルの最小受光感度を実現しております。また、70dB以上の測定ダイナミックレンジを有するため、形状、材料、色、表面状態を問わず、非接触・ 高速・高精度測定が可能です。

33

OPS-1000のシステムには、データ取得用の専用ソフトウェアが付属しています。Windows®ベースで、部品の測定に使いやすいソフトウェアです。このソフトウェアでは、
3D点群データ、3D光強度データを取得することができます。ファイル形式はPCD (Point Cloud Data) 形式です。

34

Wafer Thickness Mapping

3D Optical ProfilingHigh Accuracy and High Environmental Resistance
Wafer Thickness Mapping System with Various Measurement Parameters

Application Examples
In- Situ Inspection

Valve Body

Fresnel Lens

The TMS-2000 measures wafer flatness down toan exceptional 1 nm repeatability even inchallenging conditions for wafers up to 12
inches in diameter. Conventional wafer
thickness mapping techniques struggle withaccuracy especially when faced with unstableenvironments involving extreme temperaturechanges or vibration. TMS-2000's innovative
hardware design effectively minimizes theimpact of these environmental factors. The
high-speed measurement capability andcompact size makes it an ideal choice formeasuring flatness for a range of wafer materials
including Silicon (〜P++), SiC, Sapphire, Glass,
LiNbO3 and SOI.

Features
Metal Spring

Butterfly Package

1. High Accuracy
High acuracy measurement using interferometric detection technique (1nm repeatability)
2. Various Measurement Parameters
Analysis of Global (GFLR, GFLD, GBIR), Site (SFQR, SFQD, SBIR), Edge (ESFQR) possible

Tribology

3. High Environmental ResistanceNo temperature control or vibration countermeasures are required due to environmental durability
4. Compact Size
Small form factor suitable for multiple applications
5. High Speed
Spiral Scanning (High speed, high density)
Metal Ball

Angled Surface

Metal Bearing

Measurement Data

Fractography

Thickness Measurement
Fan

Site Analysis

Edge Analysis

Metal Pipe

TMS-2000はウエハの厚み分布を非接触で計測し、平坦度を1 nmの精度で計測可能です。本製品は環境変化にも高い耐性を有し、且つ小型、高速、低価格を実現しております。インライン検査を含め、高スループットを活かした幅広いアプリケーションへの導入が可能です。

35

36

Business Solutions

Business Solutions
santecスマートグラスソリューション

Test & Measurement
Components

お客様ごとに最適なウェアラブルデバイスソリューションをご提供
現場への革新的なDX化の実現
様々なウェアラブルデバイスと作業支援システムを組み合せることができます。
多数のスマートグラス、ARグラス、ウェアラブルカメラと複数の作業支援システムやリアルタイム映像共有システム、現場ワークフロー
システムなどから、最適な組み合わせを選択できます。
社内と現場との円滑なコミュニケーション、現場のハンズフリー作業、現場のDX化を実現することで、現場の様々な課題を解決します。
遠隔からの現場作業支援

現場ワークフローのDX化

movu

若手を配置

熟練者を配置
現場へ遠隔で作業指示・支援

紙の作業手順書を電子化

作業手順書等を作業工程ごとに表示
音声による操作で完全ハンズフリーと現場DX化を実現

OCT

本部からの支援を受けハンズフリー作業

情報ネットワーク

５

現場のDX化IT Solutions※「ライブサポーター／LIVE SUPPORTER」はサンテック株式会社の登録商標です。

LiveSupporter Web
https://www2.santec.com/ls/

37

38

Business Solutions

Business Solutions

ARグラス

スマートグラス
NaviGator 500 / 520

Magic Leap 2●●●

深度カメラ /IR レシーバ

IR アイトラッキング
LED's x 6

Magic Leap OS (Android 10 AOSP based)

●

CPU

CPU
AMD 7nm Quad-core Zen 2 X86 core (8 threads)
14 core computer vision processing engine (CVIP)
Max CPU frequency 2.4 GHz
Cache sizes: 512kB L2 per core and 4MB total L3 cacheGPU
AMD RDNA 2
2 RB+ (Render Backends)
Max Gfx frequency capped at 1.1 GHz
256GB / 16GB LPDDR5

ﾃﾞｨｽﾌﾟﾚｲ解像度 / 視野角

片目 1440 x 1760 120 Hz refresh rate20 to 2000
nits brightness / 水平 44.6 垂直 53.6 対角 70

カメラ

12.6M pixel autofocus RGB camera
動画 : 4K 30fps or 1080p 60fps

防塵防水堅牢性能

21 CFR 801.410 "Impact testing of lenses"

撮影用カメラオートフォーカス

あり

深度照明 /IR エミッタ

バッテリー※常時ﾈｯﾄﾜｰｸ経由による映像
および音声の共有を行った場合

5,776mAh
約 3.5 時間

入力インターフェース

無線コントローラ、ハンドトラキング、音声操作、
アイトラッキング、BT キーボード

センサー

6-DOF
3 つのワールドカメラ､4 つのアイトラッキングカメラ､深度カ
メラ /IR レシーバ､IR アイトラッキング LED's 6､2 磁気セン
サ､2 高度計､深度照明 /IR エミッタ､環境光センサ、4IMUs

マイク / スピーカー

4 つのマイク
2 つの立体音響スピーカー
BT 接続

環境光センサ

2 磁気センサ●OS

内部ｽﾄﾚｰｼﾞ / RAM

多彩なセンサーを搭載
フロントワールド
カメラ

両眼シースルー

4 IMUs2 高度計

サイドワールド
カメラ

サイドワールド
カメラ

アイトラッキング
カメラ

●

NEW

NaviGator Z１●●●

Navigator 500 / 520

タイプ
OS
CPU
内部ｽﾄﾚｰｼﾞ / RAM
ﾃﾞｨｽﾌﾟﾚｲ解像度 / 視野角
ディスプレイ

片眼Android 12 (AOSP) + WearHF ハンズフリーインターフェイス
Qualcomm ® QCS6490
128GB 内部ストレージ、8GB RAM
HD (1280x720) / 24
24 ビットカラー LCoS、対角 0.35 インチ、屋外でも使用可能

動画：1080p @ 60fps, >6X zoom in HD ( 手振れ補正機能搭載 )静止画：4800 万画素 ※モジュラー型 (Zone 1 の外で交換可能 )LED フラッシュライト
IP66、本質安全防爆（ATEX ゾーン 1 および CSA C1/D1） 2m 落下試験済み、MIL-STD-810Hあり

2,600mAh※約 4 〜 6 時間 ※ホットスワップ対応

2550mAh※取り外し可能な充電式（ゾーン 1 以外） 約 4 〜 6 時間

入力インターフェース
センサー

音声操作
3 軸加速度計、ジャイロスコープ、eCompass

音声操作
9-DOF (3 軸加速度センサー、地磁気センサー、ジャイロスコープ ) 手ぶれ補正ソフトウェア

GPS

GPS、Glonass、BeiDou、Galileo、QZSS、SBAS、NavlC サポート
デュアル周波数 GNSS(L1＋L5) Qualcomm センサー・アシスト・ポジショニング

Wi-Fi6重量

ヘッドセット：260g
コンピュートパック：420g ( ケーブル含めず )動作温度

10℃〜 30℃

Wi-Fi
Bluetooth
重量
動作温度

アクティブノイズキャンセリング機能付きデジタルマイク 4 個搭載
最大 100dBA の騒音下でも正確な音声認識を実現 94dBA のスピーカー内蔵
802.11 a/b/g/n/ac - 2.4 GHz and 5GHz
Bluetooth 5.1
267g / 274g
-20℃〜 55℃、相対湿度 5% 〜 95% ( 非結露 )GPS, GLONASS, NavIC, BeiDou, Galileo, QZSS, SBAS
アクティブノイズキャンセリング機能付きデジタルマイク 4 個搭載
最大 100dBA の騒音下でも正確な音声認識を実現 94dBA のスピーカー内蔵
Wi-Fi 6 (802.11ax) および Wi-Fi 6E (6 GHz)
Bluetooth Low Energy 5.2
383g-20℃〜 55℃

ウェアラブルカメラ
●
●
●

NEW

●

●

●

●

●

M4000
●
●
●

Saﬁe Pocket 2 Plus

Vuzix 製 非透過型片眼スマートグラス
ハイスペックと軽量化を両立
市場を牽引する老舗メーカーのフラッグシップモデル
どのような現場にもフィットする汎用性
業種を問わず様々な現場で活躍中
NEW

Vuzix 製 透過型片眼スマートグラス
M400 の透過型版
解像度 854 480 という大きなディスプレイ搭載

Xacti LIVE●●

Blade 2●●●

Vuzix 製 シースルー型片眼スマートグラス
超軽量かつ抜群の掛け心地
スマートグラスの中では、最も斬新的なデザイン

M400

M4000

●

Blade 2片眼シースルー
Android 11
8 Core 2.52GHz Qualcomm XR1
64GB / 6GB
854 480 / 28 度
24 ビットカラー DLP ディスプレイ 5000nits
動画：4k 30fps or 1080p 60fps
静止画：1280 万画素
あり
IP67 相当あり

片眼シースルー
Android 11
Quad Core ARM
40GB
480 480 / 19 度
ウェーブシースルーディスプレイ 輝度 :2000nitss
800 万画素なし
あり

750mAh or 3,350mAh or 4,800mAh※約 2 〜 7 時間 ※ホットスワップ対応

750mAh or 3,350mAh or 4,800mAh※約 2 〜 7 時間 ※ホットスワップ対応

約 500mAh※約 30 分〜１時間
タッチパッド､音声操作※ペアリングしたスマホによる操作可能
記載なし
あり / あり ※ノイズキャンセリング機能搭載
5.0 and 2.4GHz Wi-Fi
あり
90g

タイプ
カメラ

Saﬁe Pocket 2 Plus （セーフィー ポケット ツー プラス）
ウェアラブルカメラ

動画：100 万画素 30fps / 静止画：200 万画素

Xacti LIVE (CX-WL100)
ウェアラブルカメラ

動画：Full HD / 静止画：約 1247 万画素 水平線モード・目線モード搭載
水平 90 / 垂直 58 (VGA 時垂直 73 ) 広角モード時 対角 180

画角水平 120 ( 手ぶれ補正 ON 時 約 88 ) 垂直 86

防塵防水堅牢性能

IP67

IP67 ( カメラヘッド部 )4,200mAh※最大 8 時間

モバイルバッテリー等から給電 ※消費電力 Max.2.6W

ライト

オートフォーカス
バッテリー※常時ﾈｯﾄﾜｰｸ経由による映像
および音声の共有を行った場合
入力インターフェースGPS

白色 LED を ON/OFF 操作可能

あり / あり (Bluetooth ﾍｯﾄﾞｾｯﾄ付 )Wi-Fi / LTE

動作温度

スマートフォンより操作

物理ボタン○ (BeiDou)

LTE SIM (NTT ドコモ系 MVNO) 内蔵
Wi-Fi (LAN IEEE802.11 a/b/g/n/ac)

重量

なし

なし

マイク / スピーカー

Bluetoothなし

IT Solutions

片眼Android 11
8 Core 2.52GHz Qualcomm XR1
64GB / 6GBnHD 640 360 / 16.8 度
24 ビットカラー有機 EL ディスプレイ 2000nits
動画：4k 30fps or 1080p 60fps
静止画：1280 万画素
あり
IP67 相当
あり

物理ボタン､タッチパッド､音声操作 ※ペアリングしたスマホによる操作可能 物理ボタン､タッチパッド､音声操作 ※ペアリングしたスマホによる操作可能
3 軸ジャイロ、3 軸加速度センサー、3 軸地磁気センサー
3 軸ジャイロ、3 軸加速度センサー、3 軸地磁気センサー
あり / ありあり / あり
Wi-Fi
Wi-Fi
Bluetooth 5.0 BR/EDR/LE
Bluetooth 5.0 BR/EDR/LE
190g ( ケーブル、バッテリー、メガネフレーム込 )190g ( ケーブル、バッテリー、メガネフレーム込 )ザクティ製 約 29g の超軽量ウェアラブルカメラ＆映像配信サービス
強力なブレ補正と水平維持で映像酔いなし
ストレスフリーで眼鏡・胸・帽子などに装着
国交省の定める遠隔臨場仕様に適合

OCT●

セーフィー製 ウェアラブルクラウドカメラ
電源 ON するだけで録画、配信可能
簡単操作で現場フレンドリー
国交省の定める遠隔臨場仕様に適合

movu

●

入力インターフェースセンサー
マイク / スピーカー
Wi-Fi
重量
動作温度

Navigator Z１

LED フラッシュライト
IP66 MIL-STD-810H、6.5 フィート / 2 メートル落下試験
あり

ライト
防塵防水堅牢性能
オートフォーカス
バッテリー※常時ﾈｯﾄﾜｰｸ経由による映像
および音声の共有を行った場合

Wi-Fi

M400

タイプ
OS
CPU
内部ｽﾄﾚｰｼﾞ / RAM
ﾃﾞｨｽﾌﾟﾚｲ解像度 / 視野角
ディスプレイカメラ
ライト
防塵防水堅牢性能
オートフォーカス
バッテリー※常時ﾈｯﾄﾜｰｸ経由による映像
および音声の共有を行った場合

片眼Android 11(AOSP)＋WearHF™ハンズフリーインターフェイス＋エンタープライズ機能
2.0 GHz 8-core Qualcomm® Snapdragon™ 662 with Adreno 610 GPU - OpenGL® ES 3.2 & OpenCL™ 2.0
64GB 内部ストレージ /4GB RAM/Micro SD スロット ( 最大 512GB)
500：WVGA (854 480) / 20 520：HD (1280x720) / 24
24 ビットカラー LCD、0.32 / 0.35 インチの対角、屋外視認性
動画：〜 1080p @ 60fps, >6X zoom(HD) ( 手振れ補正機能搭載 )静止画：4800 万画素 ※望遠モード搭載

カメラ

マイク / スピーカー

スマートグラス

Realwear 製 非透過型片眼スマートグラス
前モデルより軽量化、CPU および RAM がハイスペック化
国内唯一の防爆対応のスマートグラス

Components

アイトラッキング
カメラ●

Realwear 製 非透過型片眼スマートグラス
耐ショック性を備えたタフネスモデル
アーム式ディスプレイのため、常に視界を確保可能
産業用スマートグラスとして多数の現場で導入○ ( スマートフォン側の GPS 利用可能 )あり / なし ( 映像配信時スマホのマイク / スピーカー /Bluetooth イヤホンを使用 )Bluetooth 4.2（HFP/HSP）

約 180g ( バッテリー含む )-20℃〜 50℃※「ライブサポーター／LIVE SUPPORTER」はサンテック株式会社の登録商標です。

※スマートフォン側で通信※スマートフォン側で通信

約 29.5g ( カメラヘッド部分のみ )-5℃〜 40℃

LiveSupporter Web
https://www2.santec.com/ls/

39

Test & Measurement●

Magic Leap 製 法人向けシースルー型
AR/VR ヘッドセット
圧倒的な広視野角とコンテンツの見やすさを実現
多数のセンサーと高性能 CPU を搭載
今後の XR 業界を牽引する最新デバイス

タイプ

santecNEW

40

Business Solutions

Business Solutions
リモート支援ツール

ワークスタイルの変革

All-in-one

安心・安全遠隔コントロールの定番ツール

リモート保守リモートサポート

Test & Measurement

リモートソリューション

場所や時間にとらわれず、オフィスにある自分の PC を

WEB会議

遠隔操作できる、リモートワークツール

オールインワン型のリモートソリューション

PCはもちろんスマートフォン ＆ タブレットからも簡単リモートアクセス

Online ScreenAssistは「リモートサポート」「リモート保守・メンテナンス」「WEB商談」「WEB会議」を標準装備しています。
リモート操作支援・リモート保守支援・リモートアクセスでのテレワーク支援を実現します。

Android

Components

Online ScreenViewはいつでもどこでも遠隔でPCを操作できるサービスです。
社外から社内システムの閲覧、操作を実現するとともに高度なセキュリティ機能も合わせてご提供。
不正アクセスを防ぎ、安心してご利用いただけます。
テレワークでの業務遂行を可能にし、生産性向上に貢献します。

接続イメージ

santecリモートアクセスツール

１ライセンスで使える豊富な機能
iPhoneiPad
iPad

9:45 AM

外出先
100%

姉妹サービス ISL Light
アプリ 無償 にて利用可能

オフィス

社

内

社外・自宅

お客様

社

内

社

内

お客様

自宅
中継サーバ

ネットワーク設定変更不要

AES256Bit
暗号化通信
PC(Win/Mac)

安全、便利！ 各企業にあった運用管理・機能を実現できます！
ブラックスクリーン機能※Windows professional Edition 以上が接続先に必要。

※グラフィックカードの種類によっては、ブラックスクリーン機能に対応できない場合があります。

リモート切断後自動Windowsロック

マルチモニタ対応

WEB 会議コールセンターなどの業務でお客様満足度の向上
エスカレーション機能
オペレータ ID 複数発行
ファイル・フォルダ送受信

データセンターなどの遠隔地サーバの管理バッチファイルの転送可能
異常発生時の対処可能
トラブル復旧時間の削減

リモートセールスで新たな営業手法と営業機会を創出
プレゼン実施
デモ実演
遠距離顧客を簡単フォロー

資料や画面共有ができる WEB
会議機能も標準搭載
様々な部門で利用可
1 ライセンスで 10PC まで
ダウンロード実行で簡単接続

同時接続ライセンスでコスト削減
今まで登録台数分ライセンスを購入していた場合は、同時接続ライセンスによる課金のため、コストを大幅に削減できます。また、オペレータや接続先コンピュータに台数の制限はなく無制限で登録できるため、簡単で気軽に運用ができます。

詳細設定

氏名

長谷川
有効

ショートカットキーの実行

キーボード＆マウス操作

ドラッグ＆ドロップによるファイル転送

レコーディング

画面拡大・縮小

画面カラー数

アクセスパスワードの設定

自動ポート設定

描画

アクセス履歴マーカー

ワンタイムパスワード※「オンラインスクリーンビュー／ONLINE SCREEN VIEW」はサンテック株式会社の登録商標です。

Ctrl+Alt+Delの送信

セッション再接続

電源管理
（WOL）

アクセス通知ONLINE ScreenView Web
https://www2.santec.com/osv/

41

キーボード＆マウス操作

ドラッグ＆ドロップによるファイル転送

Ctrl＋Alt＋Delの送信

ショートカットキーの実行

オペレーターID作成

レコーディング

セッション再接続

自動ポート設定

自動レコーディング

音声・ビデオチャット

アクセスパスワードの設定

デスクトップ画面共有

デスクトップ共有

再起動モードリモート保守･メンテナンス対象端末も無制限

機能一覧

機能一覧
ホワイトボード

する分だけ

IT Solutions\\santecsol\hasegawa

最大同時接続数

スクリーンショットの取得

ライセンスは同時に接続

オペレータアカウントの作成は無制限

ユーザー名

アカウント

この場合は同時接続が１なので１ライセンスで OK

ホワイトボード

テキストチャット

スクリーンショットの取得
権限管理

ブラックスクリーン
セッション履歴ワンタイムパスワード

OCT

利用者権限設定

サウンド共有機能

Microsoft RDP使用可能 (代替接続)

WEB 商談操作される側２要素認証（Mail/携帯ショートメッセージ/Google OneTimePassword)

リモート保守・メンテナンス

操作する側

端末認証

リモートサポート

movu※前提条件が必要となります。

WAKE
N
LAN

画面拡大・縮小アプリケーション共有
アクセス制限画面カラー数

プレゼンテーション
アクセス通知アクセス履歴※「オンラインスクリーンアシスト／ONLINE SCREEN ASSIST」はサンテック株式会社の登録商標です。

ONLINE ScreenAssist Web
https ://www2.santec.com/osa/

42

Business Solutions
未知のランサムウェアに対応

ある日突然データが暗号化されたら元に戻すために身代金を払いますか？
暗号化されるファイルの動きに着目

リアルタイムバックアップ復元

他社ソフトとの共存も可能

状況認識技術で未知のランサムウェアも検出・遮断

バックアップからファイルを復元。
ファイルの変更削除や毀損行為を記録。

低いPC負荷率。
動作が軽いため通常業務に影響を与えません。

AppCheckはランサムウェアそのものではなく、毀損されるファイル状況の変化に注目しています。
検知

Detection① ランサムウェア暗号化検知

対応

Response② リアルタイムバックアップ③ ランサムウェア遮断／駆除

復旧

Recover④ データを復旧

暗号化型ランサムウェア対策の決定版（クラウド管理 Web サイト）
最低 １ライセンス〜

最低 １ライセンス〜

基本 １ライセンス〜

検疫・復旧

ランサムウェア対策

フォルダ保護
ログ提供
一括ポリシー設定

中央管理機能（CMS）
オプション

エージェントロック機能
モニタリング・レポート機能
インストーラー生成機能
Edge / Chrome / Firefox

Windows 7 以降

稼働 OS ／対応ブラウザ

エージェントソフトダウンロードとアクティベー
ション時にインターネット接続必須
Windows7 は MS に起因する問題は対応できません

エージェントソフトダウンロードとアクティベー
ション時にインターネット接続必須

インターネット接続必須
オフライン端末は管理できません

各商品とも 1 年・2 年・3 年・4 年・5 年利用のご契約ライセンスをご用意しております。

AppCheckWeb
https://www2.santec.com/app/

43