■Specification tabie
ＫＴＬ‐ＰＲＯ-Models
Temperature Range

ＫＴＬ‐ＺＥＲＯ

ＫＴＬ‐ＰＲＯＫＴＬ‐ＰＲＯ‐０４

ＫＴＬ‐ＺＥＲＯ‐Ａ

200 ～ 2000 ℃

300 ～ 2000 ℃

ＫＴＬ‐ＺＥＲＯ‐Ｂ

ＫＴＬ‐ＺＥＲＯ‐Ａ‐０４

ＫＴＬ‐ＺＥＲＯ‐Ｂ‐０４

200 ～ 1200 ℃

700 ～ 3000 ℃

300 ～ 1200 ℃

Display

Inside of finder,Side panel (digital display)

Measurement modeC O N T : Continuous display
P E A K : Maximum value display
VALLEY : Minimum value display
H O L D : Retention

Accuracy

±1 ％

Ԑ

=1.00

Reproducibity±1 ℃

Ԑ=1.00Emissivty

0.10 ～ 1.00 （Digital 2 way )Response time

Sampling time 0.10 ～ 1.00 sec Variable

SensorInGaAsSi

InGaAsSi

InGaAsWavelength

1.6μm

0.9μm

1.6μm

0.9μm

1.6μm｝

RS ‐ 232C
DC 0 ～ 10 V
DC 4 ～ 20 mA

External

3 Output(Standard)

Model : Coaxial single lens reflex system
Objective : f=127 mm， F=4.2Focus adjustment range : 270 mm ～ ∞Measurement angle : 0.45℃
Finder : Viewing angle=6.2℃

Optical system

Power supply

AC 85 ～ 264 V

Power consumption

50/60 Hz (AC adaptor supply)

30 VA (AC adaptor supply)

Working temperature

0 ～ 50 ℃ (Nodew)

Weight1.4 kg (Only a main part)■KTL Appearance size(mm)■Grip/GL‐ST (mm)■SoftwareBluetooth (Option)■AC/DC Appearance size(mm)■Finder mirror(mm)■Option
SC9‐USB(232C→USB Conversion cable)

BT‐100‐USB(Bluetooth→USB Radio adapter)

Cautions: About this specification, it may change without a preliminary announcement for an improvement.

Agency26‐23 SHINKOIWA 3‐CHOME KATUSHIKA‐KU TOKYO
124‐0024 JAPANT E L 03‐3655‐0541
F A X 03‐3655‐0546
E‐mail info@leccompany.co.jp
U R L http://www.leccompany.co.jp

●The KTL series is a high-spec rediation thermometerwith fast response, wide range, and through-the-glass
measurement capability.
Capable of small spot measurement of φ1/270㎜.
(KTL-PRO-04 is φ0.4/270㎜)●Sampling speed is as fast as 0.01sec.
(Variable between 0.01 and 1.00 sec in 0.01 increments.)●2-way peak holdIn addition to the general peak hold, it has an auto-reset peak hold.
This makes it very easy to control the temperature of the line.
●3 outputs as standard equipment(RS-232C, 0～10V, 4～20mA)
Analog can be scaled to ZERO and SPAN as desired.
It can be easily connected to recorders and controllers.
●Temperature analysis software included.
windows software is includes as standard.
Data can be saved and various settings can be made from a PC.
( * This is a free software and is not supported.)

KTL

KTL-P R O (200℃ ～ 2000℃）
KTL-ZERO (ZERO-A:500℃ ～ 3000℃）
KTL-ZERO (ZERO-B:200℃ ～ 1200℃）
In futhermor, measurement of minute area
KTL-P R O-04 (or KTL-ZERO-04）

The KTL series is a high-spec radiation thermometer with
fast response, wide range, small spot measurement, andthrough-the-glass measurement capabilities.

■Option■Side control panel

Hold key(ENT)
Minmum mode key(DOWN)
Maximum mode key(UP)Continous mode key(FUNCTION)*Temperature display*Setup
RS‐232communication speed
Emissivity
Scaling■Finder viewCK-1(Carrying case)

Measurement
Point

Optical fiber
DisplayStores the KTL main unit and AC adapter■System reference diagram

GL-DX(Charger and set of grip)

KTL-ZERO/PRO φ73

φ59

φ44 φ29

φ14φ6.4 φ4.9
φ3.4 φ2.0
φ1.0
KTL-ZERO/PRO-04 φ29.2 φ23.6 φ17.6 φ11.6 φ5.6
φ2.6 φ2.0 φ1.4 φ0.8φ0.4Only for the charging battery
GL-ST(grip)
RS-232C communication

10m

8m

6m

4m

2m

1m

80cm

60cm40cm 27cm
Ｓofｔware for WindowsMonitorInterval
Graphical

Only for the alkaline battery
LA-1G(Free camera platform / gray)LM-2(Finder mirror)

Analog output DC0～10V,4～20mA

F02-002(Air purge unit)

Alarm

Meter relay(MK)LA-1B(black)Fix the KTL at any angle
SL-68G(Tripod
U-8000(Tripod/Standard) /Fine adjustment type)The grip can be mounted ona tripod without being attached

Check the focas and position bylooking
through the view finder from all directions 360°The KTL and its power supply are stored inside
the case to protect the KTL from heat and smoke*Example of the temperature control
Temperature control for chip brazing

Temperature measurement in a
Temperature control for frame brazing vacuum chamber

232→USB※Bluetooth communication

Control of high frequecy heating

Temperature management of rod

Temperature management ofinside of the furnaceControl

Digital controller

RecordRecoder■Specification tabie
ＫＴＬ‐ＰＲＯ-

ModelsTemperature Range

ＫＴＬ‐ＺＥＲＯ

ＫＴＬ‐ＰＲＯＫＴＬ‐ＰＲＯ‐０４

ＫＴＬ‐ＺＥＲＯ‐Ａ

200 ～ 2000 ℃

300 ～ 2000 ℃

ＫＴＬ‐ＺＥＲＯ‐Ｂ

ＫＴＬ‐ＺＥＲＯ‐Ａ‐０４

ＫＴＬ‐ＺＥＲＯ‐Ｂ‐０４200 ～ 1200 ℃

700 ～ 3000 ℃

300 ～ 1200 ℃

Display

Inside of finder,Side panel (digital display)

Measurement modeC O N T : Continuous display
P E A K : Maximum value display
VALLEY : Minimum value display
H O L D : Retention

Accuracy

±1 ％

Ԑ

=1.00

Reproducibity

±1 ℃

Ԑ

=1.00

Emissivty

0.10 ～ 1.00 （Digital 2 way )Response time

Sampling time 0.10 ～ 1.00 sec Variable

SensorInGaAsSi

InGaAsSiInGaAs

Wavelength

1.6μm

0.9μm

1.6μm

0.9μm

1.6μm｝

RS ‐ 232C
DC 0 ～ 10 V
DC 4 ～ 20 mA

External

3 Output(Standard)

Model : Coaxial single lens reflex system
Objective : f=127 mm， F=4.2
Focus adjustment range : 270 mm ～ ∞Measurement angle : 0.45℃
Finder : Viewing angle=6.2℃

Optical system

Power supply

AC 85 ～ 264 V

Power consumption

50/60 Hz (AC adaptor supply)

30 VA (AC adaptor supply)

Working temperature

0 ～ 50 ℃ (Nodew)

Weight

1.4 kg (Only a main part)■KTL Appearance size(mm)■Grip/GL‐ST (mm)■SoftwareBluetooth (Option)■AC/DC Appearance size(mm)■Finder mirror(mm)■Option
SC9‐USB(232C→USB Conversion cable)

BT‐100‐USB(Bluetooth→USB Radio adapter)

Cautions: About this specification, it may change without a preliminary announcement for an improvement.

Agency26‐23 SHINKOIWA 3‐CHOME KATUSHIKA‐KU TOKYO
124‐0024 JAPANT E L 03‐3655‐0541
F A X 03‐3655‐0546
E‐mail info@leccompany.co.jp
U R L http://www.leccompany.co.jp