ORDERING INFORMATIONMODEL : M2EXTFACTORY USE ONLY➠➠➠➠➠➠➠➠

PLEASE FILL IN THIS SECTION
ModelJob No.

Approved by(Sales office)Company

Ser No.

Issued by(Sales office)

Name

Sales

Approved by(Factory)
Set by(Factory)P/O No.

Ser No.
Specify the items you want to change.
Default setting will be used if not specified.
■ INPUT SETTING
FactoryITEM

SET VALUEDEFAULT VALUECOMMENTSInternalcheck

Input range

Temperature unit□（PR）
□ K（CA）□ E（CRC）□ J（IC）□ T（CC）□ B（RH）□ R□ S□ C（WRe 5-26）□ N□ U□ L□ P（Platinel Ⅱ）

K（CA）

Choose among 13 types□□ Celsius□ Fahrenheit□ AbsoluteCelsius

Choose among 3 types□

Specify within settable range in the table 1.

□

0 % input setting

0.0

100 % input setting

Input decimal point

Burnout

CJM / Fixed temp.

Filter time constant*1

MG CO., LTD.
www.mgco.jp

1000.0□ No decimal point□ 1 place of decimals□ 2 places of decimals

Choose among 3 types
1 place of decimals

Set the decimal point position for 0 %□input setting and 100 % input setting.

□ None□ Down□ UpUP

Choose among 3 types□ CJM□ Fixed terminal temperature（）

CJM

Choose among 2 types
When Fixed temp. is used, specify desired
value （-50.00 to +100.00℃）

0 sec.

Specify the range between 0 and 30 sec.

□□□

M2EXT ORDERING INFORMATION ESU-5145 Rev.4 P1/4

■ OUTPUT SETTING
FactoryITEM

SET VALUEDEFAULT VALUECOMMENTSInternalcheck□ 0 to 20 mA DC
Output range□ -5 to +5 V DC

0 to 20 mA DC

Choose among 3 types□Specify within settable range in the table 2.

□□ -10 to +10 V DC
4.000 mA
0 % output setting-5.000 V-10.000 V
20.000 mA

100 % output setting

5.000 V
10.000 V■ DISPLAY SETTING
FactoryITEM

SET VALUEDEFAULT VALUECOMMENTSInternalcheck

0 % output scaling

Specify the range between –99999 and0.00□

999999.
100 % output scaling

Specify the range between –99999 and100.00□

999999.

Output decimal point□ No decimal point□ 1 place of decimals□ 2 places of decimals□ 3 places of decimals□ 4 places of decimals□ 5 places of decimals

2 places of decimals

Choose among 6 types
Set the arbitrary decimal point position for
0 % output scaling and 100 % outputscaling.

□Choose from the table 3, or specify fromUnit (OUT Scaling)%table 4 ‘Settable characters’ within 13□

characters.
Display setting

Upper:

Upper: INPUT

Lower:

Lower: PERCENT

Brightness

4Display timeout

10 min.

Choose from the setting value in the table□5.
Specify among 1 (darkest) to 4 (brightest).
Specify the range from 0, 1 to 60 min.
Set ‘0 min.’ to display ‘always on’.

□□

*1. Filter time constantSet filter time constant of the first order lag filter. The first order lag filter is available with setting time. When ‘0’ is set to this parameter, the firstorder lag filter is not available (Response time: ≤ 0.5 sec. (0 ➝ 90 %)). The first order lag filter is equivalent to general CR filter. The settingtime constant is the time to follow until about 63 %, when input varies from 0 % to 100 %.

MG CO., LTD.
www.mgco.jp

M2EXT ORDERING INFORMATION ESU-5145 Rev.4 P2/4

Table 1MIN. SPAN
INPUT RANGE

C, K°Ｆ

SETTABLE RANGECFK（PR）

0.00 to 1760.00

32.00 to 3200.00

273.15 to 2033.15

K （CA）-270.00 to +1370.00-454.00 to +2498.00

3.15 to 1643.15

E （CRC）-270.00 to +1000.00-454.00 to +1832.00

3.15 to 1273.15

J （IC）-210.00 to +1200.00-346.00 to +2192.00

63.15 to 1473.15

T （CC）-270.00 to +400.00-454.00 to +752.00

3.15 to 673.15

0.00 to 1820.00-32.00 to +3308.00

273.15 to 2093.15-50.00 to +1760.00-58.00 to +3200.00

223.15 to 2033.15-50.00 to +1760.00-58.00 to +3200.00

223.15 to 2033.15

C （WRe 5-26）

0.00 to 2315.00

32.00 to 4199.00

273.15 to 2588.15

N-270.00 to +1300.00-454.00 to +2372.00

3.15 to 1573.15

U-200.00 to +600.00-328.00 to +1112.00

73.15 to 873.15

L-200.00 to +900.00-328.00 to +1652.00

73.15 to 1173.15

0.00 to 1395.00

32.00 to 2543.00

273.15 to 1668.15

B （RH）
R20.0036.00

S

P （Platinel II）

Note. For set value, specify so that value is within the settable range depending on type of the T/C.

Table 2OUTPUT RANGECurrent output
Voltage output

MIN. SPANSETTABLE RANGE

0 to 20 mA DC

1.000 mA

0.000 to 20.000 mA-5 to +5 mV DC

0.250 V

-5.000 to +5.000 V-10 to +10 V DC

1.000 V

-10.000 to +10.000 V

Table 3AVAILABLE UNITS
DC, AC, mV, V, kV, μA, mA, A, kA, mW, W, kW, var, kvar, Mvar, VA, Hz, Ω, kΩ, MΩ,cm, mm, m, m/sec, mm/min, cm/min, m/min, m/h, m/s2, inch, L,L/s, L/min, L/h, m3, m3/sec, m3/min,m3/h, Nm3/h, N∙m, N/m2, g, kg, kg/h, N, kN, Pa, kPa, Mpa, t, t/h,°C, °F, K, %RH, J, kJ, MJ, rpm, sec, min, min-1, pH, %, ppm, deg, (blank),

Table 4
SETTABLE CHARACTERS
0 − 9 A − Z a − z ! ” # $ % &’ ( ) = - +  ^ |@`[]{};:<>?_,./MG CO., LTD.
www.mgco.jp

M2EXT ORDERING INFORMATION ESU-5145 Rev.4 P3/4

Table 5Upper
SETTING VALUEDESCRIPTION

INPUT

Input engineering unit value

VOLTAGE

Input electromotive force value

PERCENTPercent value*2

OUTPUT

Output engineering unit value

OUTPUT (Scaling)

Output scaling

Lower
SETTING VALUEDESCRIPTION

INPUT

Input engineering unit value

VOLTAGE

Input electromotive force value

PERCENT

Percent value*2

OUTPUT

Output engineering unit value

OUTPUT (Scaling)

Output scaling

None

No display*2. Input percent value.

MG CO., LTD.
www.mgco.jp

M2EXT ORDERING INFORMATION ESU-5145 Rev.4 P4/4