Pressure transmitter

Relative pressure transmitter
Type 519

The pressure transmitter type 519 has a compact and robust
design and a very high measurement accuracy.
The flush diaphragm offers the use in the process technique ofpasty media.
Beside various pressure- and electrical connectors there arepressure ranges from 400 mbar to 60 bar full scale available.
This sensor utilises a ceramic technology, developed by Huba
Control and for the last 20 years, in millions of applications.

Pressure range
0 ... 0.4 – 60 bar+ Flush diaphragm+ High measurement accuracy+ Available with integrated temperature measurement+ There are low pressure ranges from 400 mbar full scale
available+ Compact and rugged construction

Pressure and flow

Huba Control Type 519 | Technical data subject to change | Edition 09/2020

1/5

Technical Overview
Pressure range
Relative

0 ... 0.4 – 60 bar

Operating conditions
MediumTemperature
Tolerable overload / rupture pressure
Materials
CoverMaterials in contact with medium
Electrical overview
2 wire3 wire4 wire(with temperature)
Polarity reversal protection
Insulation voltage

Response time
Load cycle
Electrical connection
Connector DIN EN 175301-803-A
Connector DIN EN 175301-803-C (Industrial standard 9.4 mm)
Connector M12x1

Outside thread

Liquids, pasty media and gases-25 ... +125 ºC-25 ... +85 ºC-40 ... +100 ºC
max. 4.8 bar
3x FS

Pressure connection
SensorSealing material

Stainless steel 1.4404 / AISI 316L
Stainless steel 1.4404 / AISI 316L
Ceramic Al2O3 (99.6%)
FPM, EPDM

OutputPower supplyLoad
10 V
4 . .. 20 mA
10 ... 30 VDC< Supply voltage[Ohm]0.02 A
0 ... 5 V
7 ... 33 VDC>10 kOhm / < 100 nF
0 ... 10 V
12 ... 33 VDC>10 kOhm / < 100 nF
ratiom. 10 ... 90%
5 VDC ±10%
>10 kOhm / < 100 nF
Pressureratiom. 10 ... 90%
5 VDC ±10%
>10 kOhm / < 100 nF
Temperature (-40 ... +150 ºC)
ratiom. 10 ... 90%
5 VDC ±10%
>10 kOhm / < 100 nF
Short circuit proof and protected against polarity reversal. Each connection is protected against crossover up to max. supply voltage.
1000 VDC

Dynamic response
Start-up time

Pressure connection

MediumAmbientStorage< 1.6 bar
> 1.6 bar

G½G½
G¾

Pressure
Temperature T90< 100 ms
< 5 ms< 5s< 100 Hz

Protection standard
IP 65
IP 65
IP 67

Protection classIIIIIIIII

Current consumption< 23 mA< 7 mA< 7 mA< 5 mA< 5 mA< 5 mA

sealed at back DIN 3852-E with profile seal ring in FPM (-30 ... +135 ºC)
dual sealded with profile seal ring in FPM (-30 ... +135 ºC)
dual sealed with profile seal ring in FPM (-30 ... +135 ºC)

Installation arrangement
Unrestricted
Tests / Admissions
Electromagnetic compatibility
Shock acc. IEC 60068-2-27
Vibration nach IEC 60068-2-6
Explosion protection

without zone separation
with zone separation

UL

CE conformity acc. EN 61326-2-3
100 g, 11ms, half sine wave, 1x, all 6 directions
40 g, 6 ms, half sine wave, 1000x, all 6 directions
10 g, 10 ... 2000 Hz, 1 Octave/min., all 3 directions, 50 constant load
Ex II 1 G Ex ia IIC T4 Ga
Ex II 1 D Ex ia IIIC T120 ºC Da
Ex II ½ G Ex ia IIC T4 Ga/Gb
Ex II ½ D Ex ia IIIC T120 ºC Da/Db
ANSI/UL 61010-1 acc. E325110

Weight110 ... 140 g
Packaging (Please state on order)
Single packaging in cardboardMultiple packaging in cardboard (25 pcs)

accessories integrated

AccuracyStandard

Higher accuracy

Parameter pressure (> 1 bar)

Unit

Parameter pressure (> 1 bar)

Unit

Characteristic line 1) (at 25 ºC)
Resolution
Long term stability acc. IEC 60770-1
Temperature error (at -25 ºC ... +125 ºC) 2)% fs ± 0.3% fs < 0.1% fs ± 0.25
% fs/10K

Characteristic line 1) (at 25 ºC)
Resolution
Long term stability acc. IEC 60770-1
Total error band (at -25 ºC ... +125 ºC)% fs ± 0.3% fs < 0.1% fs ± 0.25% fs ± 0.5.

.

Parameter pressure (< 1 bar)

Unit

Parameter pressure ( < 1 bar)

Unit

Characteristic line 1) (at 25 ºC)
Resolution
Long term stability acc. IEC 60770-1
Temperature error (at -25 ºC ... +125 ºC) 2)% fs ± 0.5% fs < 0.1% fs ± 0.25
% fs/10K

Characteristic line 1) (at 25 ºC)
Resolution
Long term stability acc. IEC 60770-1
Total error band (at -25 ºC ... +125 ºC)% fs ± 0.5% fs < 0.1% fs ± 0.25
% fs ± 0.71)± 0.15

Parameter temperature

UnitCharacteristic lineResolutionºCºC

Test conditions:
Balance position:

2/5± 0.15.

± 3.0
0.2

45% rFvertical, pressure connection downincl. zero point, full scale, linearity, hysteresis and repeatability

Pressure and flow

Huba Control Type 519 | Technical data subject to change | Edition 09/2020

1

Order code selection table in bar
Pressure mode

Pressure range 1)

Sealing material
Version

Output / power supply

Electical connection 4)

Pressure connection 1)
Pressure range variation(optional)

29D

2211
11
1
3
3
3
3
4

Sealing material
Version

Output / power supply

Electical connection 4)

Pressure connection 1)
Pressure range variation(optional)

1)57812457012306789100
10441234
751,21,31,31,333M
12000

1231
11

Indicate W and state range on order (e.g. W0... + 0.5bar/OUT0.5...4.5V)

W
2

3

4

5

6

79DAAB
B
B
B
BCCCCD

78
124
570
1230

910110
10,20,20,20,20,20,30,30,30,30,30,30,302344123475
1,2

1,3
1,3

1,333M
12

Indicate W and state range on order (e.g. W0... + 0.5bar/OUT0.5...4.5V)

Pressure and flow

8519. X X X X X X X X X X X

Relative
Relative with higher accuracy
0 ... 6 psi
0 ... 10 psi
0 ... 15 psi
0 ... 20 psi
0 ... 30 psi
0 ... 60 psi
0 ... 100 psi
0 ... 150 psi
0 ... 200 psi
0 ... 300 psi
0 ... 500 psi
0 ... 750 psi
FPM
Fluoro elastomer
EPDM
Ethylene propylene
standardwith Ex-Admission 2)
without zone separation
Ex II 1 GD
with Ex-Admission 2)
with zone separation 3)
Ex II 1/2 GD
0 ... 5 V
7 ... 33 VDC
0 ... 10 V
12 ... 33 VDC
4 ... 20 mA
10 ... 30 VDC
10 ... 30 VDC
4 ... 20 mA
ratiom. 10 ... 90%
5 VDC ±10%ratiom. 10 ... 90%
5 VDC ±10% with temperature
M12x1 2L: IN=1 / OUT=3
3L: IN=1 / OUT=4 / GND=3 / Opt. T=2
M12x1
3L: IN=1 / OUT=3 / GND=4Connector
DIN EN 175301-803-A
DIN EN 175301-803-C
G ½ sealed at back DIN 3852-E
Outside thread
G ½ dual sealded
G ¾ dual sealded

Other pressure ranges on request

110,20,20,20,20,20,30,30,30,30,30,30,3231Pressure range 1)4519. X X X X X X X X X X X

Relative
Relative with higher accuracy
0 ... 0.4 bar
0 ... 0.6 bar
0 ... 1.0 bar
0 ... 1.6 bar
0 ... 2.5 bar
0 ... 4.0 bar
0 ... 6.0 bar
0 ... 10.0 bar
0 ... 16.0 bar
0 ... 25.0 bar
0 ... 40.0 bar
0 ... 60.0 bar
FPMFluoro elastomer
EPDM
Ethylene propylene
standard
with Ex-Admission 2)
without zone separation
Ex II 1 GD
with Ex-Admission 2)
with zone separation 3)
Ex II 1/2 GD
0 ... 5 V
7 ... 33 VDC
0 ... 10 V
12 ... 33 VDC
4 ... 20 mA
10 ... 30 VDC
10 ... 30 VDC
4 ... 20 mA
ratiom. 10 ... 90%
5 VDC ±10%ratiom. 10 ... 90%
5 VDC ±10% with temperature
M12x1 2L: IN=1 / OUT=3
3L: IN=1 / OUT=4 / GND=3 / Opt. T=2
M12x1
3L: IN=1 / OUT=3 / GND=4Connector
DIN EN 175301-803-A
DIN EN 175301-803-CG ½ sealed at back DIN 3852-E
Outside thread
G ½ dual sealded
G ¾ dual sealded

Order code selection table in psi
Pressure mode32)

without UL-Admisson3)

000

1
2
3

1
11
W

Turn-down versions max. > 1 bar2)

Delivery without female connector

Huba Control Type 519 | Technical data subject to change | Edition 09/2020

3/5

Accessories (supplied loose)

Order number

Female connector DIN EN 175301-803-A with seal
Connector DIN EN 175301-803-C with seal
Corner-wire box for connector M12x1
Corner-wire box for connector M12x1 with cable 2.0 m
Straight-wire box for connector M12x1
Straight-wire box for connector M12x1 with cable 2.0 m
Calibration certificate

103510104244106975114604114570114605104551

Dimensions in mm / Electrical connections
L1 = 54.5

L2 = 22.9

2 wire2725 Nm
Code Nr.: 1

 ,1  287 
L2 = 26.5

 ,1  287 3 wire

L1 = 57.5

L1L22725 Nm
Code Nr.: 2

 ,1  *1'  287

 ,1  287  *1'L2 = 28.1
L1 = 54.5

 ,1  287  *1'3430 Nm
Code Nr.: 34 wire

Device design with explosion protection: 4 ... 20 mA
The grounding connection is conductively connected to the pressuretransmitter housing.

4/5

connector DINEN 175301-803-A

connector M12x1

1 (IN) 2 (OUT)

1 (IN) 3 (OUT) 4 ( )Pressure and flow ,1  2877  *1'  2873Huba Control Type 519 | Technical data subject to change | Edition 09/2020

www.hubacontrol.com