Contents
p. 2Applic.
pp. 4-8

Selection
pp. 9-11

GenRadproducts
pp. 50-87

Products
pp. 12-87

HARS-L • HARS-LX • Series

Laboratory Standard
ラボ用可変標準抵抗器

Index
p. 89

Decade Resistance
Substituterp. 1 of 2IET
Labs製品で最も精密な可変抵抗器で正確Highestaccuracy version of the IET Labs resis
tance
substituters for the most exacting
さを必要とする校正あるいはテストでの応用
calibration and test applications. The HARS-LXに適しています。HARS-LXシリーズはオプ
Series features a continuous rheostat as an opションとして連続的な加減抵抗器の機能を備tion.
えています。

＊密封された低インダクタンス抵抗を使用
Hermetically sealed, low inductance resistors＊精密に固定された最小抵抗値
Precise fixed minimum resistance＊確度:20ppm - 20 ppm
High accuracy＊高安定度:
5ppm/年High stability- 5 ppm/yr＊低温度係数:
3ppm/℃
Low temperature coefficient - as low as 3 ppm/°C＊高性能な純銀接点のスイッチHigh performance solid silver contact switches＊抵抗値: 10mΩから121MΩ以上
Resistance from 10 mΩ to over 121 MΩ
＊分解能: 1mΩ、オプションで20μオーム
1 mΩ or optional 20 μΩ resolution
HARS-LX Laboratory Standard Decade Resistance Substituter(shown with optional rheostat)（上記写真のモデルはオプションの加減抵抗器付）

標準モデルMODELSSTANDARDSeriesシリーズ

HARS-L

HARS-LXHARS-Xより高性能で、コストパフォーマンスが要
Tighttolerance versions of the IET labs HARS-X
Seriesfor applications requiring a cost effective high
求される応用での標準可変抵抗器として、研究所あperformance
resistance decade standard suitable
るいは現場での校正に適しています。
for laboratory and field calibrations.

Description
内容Resistance wire for 0.1 steps and under; hermetically sealed,wirewound non-inductive resistors for 1 Ω steps and over.

Resistortype
抵抗タイプ
範 囲Range

Highestperformance decade resistance substituter
最もパフォーマンスの良い可変抵抗器で確度、安withthe tightest tolerance, stability, repeatability, and定度、再現性、温度係数のいずれも優れています。perature coefficient.
tem

10 mΩ up to 12.1 MΩ in 1 to 10 decades; (minimum may be lower for units with fewer decades)

分 解 能Resolution

1 mΩ discrete steps; 20 μΩ continuous resolution rheostat; 10 mΩ full scale, option RH.
<±(20 ppm + 0.5 mΩ); at 23°C, no zero subtraction required,
4-terminal, “true-ohm” measurement, SI traceable

InitialAccuracy (absolute)
初期確度（絶対値）<± (25 ppm + 0.5 mΩ); at 23°C, no zero subtraction required,4-terminal, “true-ohm” measurement, SI traceable.

InitialAdjustment Accuracy
初期調整確度±1 ppm for 10 kΩ steps; ±1.5 ppm for 100 kΩ steps; ±3 ppm for 1 MΩ steps.
(For increased accuracy of the 1 Ω to 1 MΩ decades, individual resistors for these decades are trimmable.)

温度係数 Coefficient
Temperature<± 5 ppm/°C for 100 Ω steps and over.

<± 20 ppm/°C for 10 Ω steps and under;<± 50 μΩ/°C for wiring and switch resistance.

安定度
Stability<± 20 ppm/°C for 1 Ω steps and under;<±15 ppm/°C for 10 Ω steps;<± 3 ppm/°C for 100 Ω steps and over;<± 50 μΩ/°C for wiring and switch resistance.

<±(20 ppm + 0. 5 mΩ)/year; <±5 ppm/year, typical.

最小抵抗値
Minimum Resistance
最大パワーPower Maximum

10 mΩ ± 0.5 mΩ; limited by the lowest settable position, “1”, of the 10 mΩ/step decade.
0.5 W per step up to 3 W total or 2 A max.

1 W per step up to 5 W total or 2 A max.

Calibration Conditions
校正条件

Four-terminal measurement, low power, at 23°C; 30% to 60% RH.

Switch Type
スイッチタイプ

11 positions, "0"-"10", multiple solid silver alloy contacts,with short term contact resistance repeatability of <100 μΩ.

破壊電圧Voltage
Breakdown
Power Coefficient

パワー係数

1500 V peak to case<±1000 ppm/W for 0.1 Ω steps and under;<±400 ppm/W for 1 Ω steps;<±300 ppm/W for 10 Ω steps;<±100 pm/W for 100 Ω steps and over.
<+50 μΩ/W for wiring and switch resistance.

IET LABS, INC. in the GenRad Tradition
534 Main Street, Westbury, NY 11590
IET cat/HARSLX p1/02-24-06

www.ietlabs.com
TEL: (516) 334-5959 • (800) 899-8438 • FAX: (516) 334-5988
17

Contents
p. 2Applic.
pp. 4-8

Selection
pp. 9-11

Products
pp. 12-87

GenRadproducts
pp. 50-87

HARS-L • HARS-LX • Series

Laboratory Standard
ラボ用可変標準抵抗器

Index
p. 89

Decade Resistance
Substituterp. 2 of 2

端子： 低サーマルEMFのベリリウム銅合金製のパインディング端Terminals:
Low thermal emf beryllium copper binding posts
子、標準の3/4インチスペース間隔のシールド端子。
with standard
3/4 inch spacing plus shield terminal; connection
from the ROオプションで機器の背面からの接続も可能。
rear of the instrument is available as option RO.

モデル
Model

Dimensions

Weight6 and 7
decades43.9 cm W x 8.9 cm H x 10.2 cm D

2.2 kg(4.8 lb)

Modelモデル

Dimensions

Weight1 decades

7.7 cm W x 7.7 cm H x 8.4 cm D
(3" x 3" x 3.3")

0.45 kg(1.0 lb)

8 decades(17.3" x 3.5" x 4")

5.1 kg(13.0 lb)

1.7 kg(3.8 lb)

9 and 10decades

48.3 cm W x 17.8 cm H x 19.7 cm D
(19.0" x 7.0" 7.8")

5.1 kg(13.0 lb)

2.0 kg(4.3 lb)

11 decades

48.3 cm W x 32.5 cm H x 27.0 cm D
(19.0" x 12.8" 10.5")

9.1 kg(20.0 lb)

2-4
decades37.5 cm W x 8.9 cm H x 10.2 cm D

5 decades(14.8" x 3.5" x 4")

オプション：
加減抵抗器
OPTIONAL RHEOSTATDISCRETEDECADESHI CURRENTLO CURRENTRHEOSTATLO SENSE

HI SENSE

オプションの加減抵抗器の構成
HARS-LX with
Optional Rheostat Configuration

高い分解能が必要なアプリケーションでは、最下段のステップに10mΩの加減抵抗器を加えると良いでしょう。加減抵抗器の分wiper is in the low potential circuit, which is the high impedanceForhigh resolution applications, a 10 mΩ rheostat may be
解能は20μΩです。接触抵抗とサーマルＥＭＦを取除くためにHARS-LXは加減抵抗器を上記に示すように構成しています。こlead. As a result, voltage and contact resistance effects areaddedfor the lowest step. It is a 20 μΩ resolution “decade”.の方法では、摺動子は高インピーダンスのリードを持つローポテンシャル回路にあります。この結果、電圧と接触抵抗の影響
removed by being effectively added to the input impedance ofInorder to eliminate contact resistance and thermal emf, theは、測定計器の入力インピーダンスに効果的に足されることにより、取り除かれます。
the measuring instrument.
HARS-LX
integrates the rheostat as shown. In this way, theORDERING INFORMATION
オーダー情報Modelモデル(Select L or LX
accuracy grade)

Total
合計抵抗値
Resistance
(Ω)

No.
of
ﾃﾞｨｹｰﾄﾞ数
Decades

Resolution
分解能(Ω)HARS-L(LX)-1-0.001
HARS-L(LX)-1-0.01
HARS-L(LX)-1-0.1
HARS-L(LX)-1-1
HARS-L(LX)-1-10
HARS-L(LX)-1-100
HARS-L(LX)-1-1K
HARS-L(LX)-1-10K
HARS-L(LX)-1-100K
HARS-L(LX)-1-1M

0.01
0.1
110100
1k
10 k
100 k
1M
10 M

1111111
1
1
10.0010.01
0.1110100
1k
10 k
100 k
1M

HARS-L(LX)-2-0.001
HARS-L(LX)-2-0.01
HARS-L(LX)-2-0.1
HARS-L(LX)-2-1
HARS-L(LX)-2-10
HARS-L(LX)-2-100
HARS-L(LX)-2-1K
HARS-L(LX)-2-10K
HARS-L(LX)-2-100K

0.11
1.111110
1.1 k
11 k
110 k
1.1 MΩ
11 MΩ

222222222

0.001
0.01
0.1
110100
1k
10 k
100 k

HARS-L(LX)-3-0.001
HARS-L(LX)-3-0.01
HARS-L(LX)-3-0.1
HARS-L(LX)-3-1
HARS-L(LX)-3-10
HARS-L(LX)-3-100
HARS-L(LX)-3-1K
HARS-L(LX)-3-10K

1.11
11.1
111
1.11 k
11.1 k111 k1.11 M
11.1 M3333
3
3
3
30.001
0.01
0.1
1
10
100
1k
10 k

OPTIONS- RH 10 mΩ rheostat for lowest decades, 20 μΩ resolution.
- RO Rear output binding posts

モデルModel(Select L or LXaccuracy grade)

Total
合計抵抗値
Resistance

HARS-L(LX)-4-0.001
HARS-L(LX)-4-0.01
HARS-L(LX)-4-0.1
HARS-L(LX)-4-1
HARS-L(LX)-4-10
HARS-L(LX)-4-100
HARS-L(LX)-4-1K
HARS-L(LX)-5-0.001
HARS-L(LX)-5-0.01
HARS-L(LX)-5-0.1
HARS-L(LX)-5-1
HARS-L(LX)-5-10
HARS-L(LX)-5-100
HARS-L(LX)-6-0.001
HARS-L(LX)-6-0.01
HARS-L(LX)-6-0.1
HARS-L(LX)-6-1
HARS-L(LX)-6-10
HARS-L(LX)-7-0.001
HARS-L(LX)-7-0.01
HARS-L(LX)-7-0.1
HARS-L(LX)-7-1
HARS-L(LX)-8-0.001
HARS-L(LX)-8-0.01
HARS-L(LX)-8-0.1
HARS-L(LX)-9-0.001
HARS-L(LX)-9-K-RM
HARS-L(LX)-9-0.01
HARS-L(LX)-10-0.001
HARS-L(LX)-11-0.001

IET LABS, INC. in the GenRad Tradition
534 Main Street, Westbury, NY 11590
18Resolution
分解能(Ω)No.
ﾃﾞｨｹｰﾄﾞof数
Decades

11.11
111.1
1.111 k11.11 k111.1 k1.111 M
11.11 M
111.111.111 1 k
11.111 k
111.11 k
1.111 1 M
11.111 M
1.111 11 k
11.1111 k
11.111 k
1.111 11 M
11.111 1 M
11.111 11 k
111.111 1 k
1.111 111 M
11.111 11 M
111.111 11 k
1.211 111 1 M
1.211 111 M
1.211 111 11 M
1.211 111 11 M
12.111 111 1 M
12.111 111 11 M
121.111 111 11 M

444
4
4
4
4
5
5
5
5
5
5
6
6
6
6
6
7
7
7
7
888
999
10
11

0.001
0.01
0.1
1
10100
1k
0.001
0.01
0.1
1
10
100
0.001
0.01
0.1
1
10
0.001
0.01
0.1
1
0.001
0.01
0.1
0.001
0.001
0.01
0.001
0.001

(Ω)

www.ietlabs.com
TEL: (516) 334-5959 • (800) 899-8438 • FAX: (516) 334-5988
Electronic cat, pp 1-45/p5/04-21-99;33