LINE SEIKI
EA-1000

DIGITAL ANEMOMETER
INSTRUCTION MANUAL

We thank you for purchasing our company’s anemometer. Aside frombeing economical and portable, this product is a high quality, durable andeasy to operate digital meter. The vane method employed is an excellent andeffective technique for measurements capable of obtaining accurate readings
even in remote locations.
We ask you to carefully read and understand this manual before using toguide you on how to gain the outmost benefits of all its function and uses.

3. Battery Replacement
1. When the upper left corner of the LCD display shows “LO BAT” , the battery
needs to be replaced.
2. In replacing the battery, slide out the battery cover until it is removed,
then take the battery out.
3. Replace the battery with a new one and position this new battery
correctly inside the meter’s battery compartment.
4. Fasten back the cover.
* The instrument cannot measure accurately after low battery indicator isdisplayed. Please replace the battery immediately.

4. Cleaning
Wipe off dust, and other dirt on the unit using a dry cloth.

Please keep this manual for future reference.

LINE SEIKI CO., LTD.

Caution·Please do not wipe the unit with benzene or thinner, this may cause
the unit to easily break and produce white marks on the chasis.

SPECIFICATIONSSAFETY INDICATION
Please strictly observe the safety instructions during the use of this product.
Read the precautionary instructions written next to the symbol indicating thecaution mark of this manual.

Caution· Please do not drop or subject this to a strong shock.
· Do not immerse this unit in water or any liquids.
This product is not water proofed.
· Do not leave this instrument inside cars, buses and other vehicles
during warm days.
· Do not expose the product into dust, high temperature and humidity,
or direct sunlight.
· Turn off the power when the product is not being used.
If the product is not to be used for a long time, please remove thebattery.
· Avoid disassembling the product.
If product failure occurs, ask for technical assistance.

1. General Specifications
Measuring Method
Display
Measuring Functions
Sampling TimeOperating Conditions
Power Supply
Battery Life
Dimensions
Accessories: Vane Method: 13 mm, LCD, 3 ½ digits display: m/s, km/h selection:
Approximately 0.4 seconds: 0 - 50ºC, less than 80% RH: 006P, MN1604 (PP3) or equivalent ... 1 piece: Approximately 90 hours: Main Unit ... 131 (H) x 70 (W) x 25 (D) mm
Sensor Head ... Diameter nearly 72 mm
Nearly 270 grams (battery and sensor included): Instruction Manual ... 1 piece
Battery ... 1 piece: V

2. Electrical Specifications
Function
m/s
km/h

Measuring Range

Resolution

0.8 - 30.0 m/s
2.8 - 108 km/h

0.1 m/s
0.1 km/h

Accuracy± (3% + 2d)± (3% + 2d)* Specifications are tested under the environment of RF field strength less than 3 V/m and frequency less than 30 MHz.

PROCEDURES
1. Front Panel Description

156POWEROFFONRANGE

EA-1000RANGE
m/s
0.8~30.0km/h

721

2.8~108

3m/s

2km/h

3A nemometer4LINE SEIKI

564

7

Main Unit

Display
Power SupplySwitch
RangeSelect Switch
BatteryCompartment
DirectionSticker
SensorSensor Handle

SensorHead

2. Measuring Procedure
1. Switch on the power.
2. Select the measuring range depending on the speed of the air to measure3. Hold the sensor head and direct it to the measuring airflow.
Get reading when display stabilizes* Measuring Consideration

The sensor part indicates the proper direction of airflow when measuring.
In doing so, allow air to run from the side of the sensor bearing the yellowsticker.

LINE SEIKI CO., LTD.
37-7, 2-chome, Chuo-cho, Meguro-ku, Tokyo, 152-0001 JAPAN
Telephone : 03-5723-891
Fax : 03-3710-4552
E-mail : webtrade@line.co.jp
URL : http://www.lineseiki.com
This manual was last revised Nov. 29, 2010. 4LU011A*Subject to change without prior notice.
All Rights Reserved,Copyright © 2010,LINE SEIKI CO.,LTD.