LINE SEIKI

G36

SELF POWERED TOTALIZER / RATEMETERLARGE LCD DISPLAY (11mm Digit Size)
10kHz Count Speed
PRESCALE FEATUREDECIMAL POINT POSITION SETTINGFREE WRITE FEATURE
DUST & SPLASH PROTECTION FRONT PANEL
MODELSMODELTYPEINPUT METHODINPUT TYPE

Count Up

Contact / Open Collector / Voltage

90° Quadrature

Voltage

G36 - 101
G36 - 102

Totalizer

G36 - 103PRESCALENO
YES

8Contact / Open Collector

Count Up/Down

G36 - 104

NO. OF DIGITS

YES

Voltage

YESv1

G36 - 201

Ratemeter

1 Input

Contact / Open Collector / Voltage

4/5YESG36 - 301

Totalizer /Ratemeter1 Input

Contact / Open Collector / Voltage

Totalizer : 8Ratemeter : 4/5v1

YES

SPECIFICATIONS
MODEL

G36 - 101

G36 - 102TYPE
DISPLAY

G36 - 103

G36 - 104

Totalizer

Add

Add/Subtract
90° Quadrature

INPUT TYPE

Contact/Open
Collector/Voltage

Voltage

Ratemeter

Totalizer / Ratemeter
Totalizer : 8 Dig. LCD

Add/Subtract(Individual input)
Contact/
Open Collector

G36 - 301

4/5 Digits LCDv1 Ratemeter: 4/5v1 LCD

8 Digits LCD (7 digits for G36-102~104 when count value is negative)

COUNT MODE

G36 - 201RateMeasurement

Add /Rate Measurement

Contact/Open Collector/Voltage

Voltage

INPUT SIGNAL

Contact : Relay, Microswitch, Open Collector

Voltage : L: 0~0.5VDC / H:2~15VDC

COUNT SPEED

Contact/Open Collector : 25Hz

Voltage : 10kHz

PULSE WIDTH

Contact/Open Collector : On/Off 20ms

Voltage : On/Off 50µs

RESET

Front Reset (push button), Remote Reset

Front Reset (push button), Remote Reset

BATTERY LIFE

5 years (at 25°C continuous)

POWER SUPPLY

Self Powered (3V Lithium Battery) (Matsushita brand BR2/3A or Fuji Denki CR2/38L(N2) recommended)v2

PRESCALE0.0001 ~ 100.0000

0.001 ~ 9999

Totalizer : 0.0001~100.0000
Ratemeter: 0.001~9999

DECIMALPOSITION

4 positions

3 positions

Totalizer : 4 positions
Ratemeter: 3 positions

RATEMETERSPECIFICATION

Accuracy: 60.2%
Refresh rate: 0.7s min Zero Time: 10s

CONNECTION

M3 Terminal Screw

PROTECTION

Front Panel: Dust proof and Splash proof (when mounted with gasket)

OPERATING
TEMPERATURE-5 ~ +40°C (non-freezing)

OPERATINGHUMIDITY

45 ~ 85% RH (non-condensing)

WEIGHT

approx. 65g

v1 Normal programming of the unit shows 4 digits of calculated value. However, when rate is set at x10, display shows 5 digits with fixed “0”.
v2 Replacement batteries are available for sale from Line Seiki.

OPERATING PROCEDURETOTALIZER

G36 - 101, 102, 103, 104
G36-101 Without Prescale
The front reset button comesdisabled from the factory. To
enable the front reset button,
install a jumper wire between
Terminals n and j.

G36-102,103,104 With Prescale
To enter Program Mode, connect
Terminals n and j using a jumper
wire. After program settings areentered, disconnect the jumper
wire to save settings and displaywill go back to Count Mode.

BUTTON FUNCTIONS (For models G36-102,103,104 only)
BUTTONRSTPROGRAM MODE

COUNT MODE

This button is used to move the cursor from one digit toanother.
The active digit is indicated by a blinking display.

NOT APPLICABLE

This button is used to set the desired value of each digit.
Keeping this button depressed will cause numbers toautoscroll.
This button is also used to set other program settings like
Front Reset enable/disable and Decimal Point Position.

RESET COUNT VALUE

By pressing RSTbutton while keeping thebutton depressed, the display will scroll from one program
screen to another. The order of scroll is as shown:
Prescale Factor setting

NOT APPLICABLE

Decimal Point Position setting
RSTFree Write Value setting
Front Reset Enable/Disable setting

PROGRAMMING PROCEDURE (For models G36-102,103,104 only)
PROGRAMPRESCALE
SETTING

DECIMAL POINT
POSITION SETTINGFREE WRITESETTINGFRONT RESETSETTINGPROGRAM SCREENSETTING PROCEDURE
When the display shows the Prescale Factor setting screen, the right most digit willblink. Choose the digit to set by pressing thebutton. The active digitwill blink.
Set the desired value of the chosen digit by pressing the RSTbutton.
Depressing the RST
button will cause the value to auto scroll.
When the display shows the Decimal Point Position setting screen, press theRST
button until the desired decimal point position is reached.

Free Write allows the counter to reset to a Free Write value other than zero. Whenthe display shows the Free Write setting screen, the right most digit will blink.
Choose the digit to set by pressing thebutton. The active digit will blink.
Set the desired value of the chosen digit by pressing the RST
button.
Depressing the RST
button will cause the value to auto scroll.
When the display shows the Front Reset setting screen, choose the desired settingof the Front Reset by pressing the RST
button. A display indicates anenabled front reset. A
display indicates a disabled front reset.
Note: The remote reset is still active even if front reset setting is disabled.

OPERATING PROCEDURE
RATEMETER

G36 - 201
To enter Program Mode, connect Terminals n andj using a jumper wire. After program settings areentered, disconnect the jumper wire to savesettings and display will go back to Count Mode.

BUTTON FUNCTIONSBUTTONPROGRAM MODE

COUNT MODE

This button is used to move the cursor from one digit toanother.
The active digit is indicated by a blinking display.

NOT APPLICABLE

This button is used to set the desired value of each digit.
Keeping this button depressed will cause numbers toautoscroll.
This button is also used to set other program settings like
Decimal Point Position and Rate Display Multiplier.

NOT APPLICABLE

By pressing
button while keeping thebutton depressed, the display will scroll from one program
screen to another. The order of scroll is as shown:Prescale Factor setting

NOT APPLICABLE

Decimal Point Position setting
Rate x10 setting

PROGRAMMING PROCEDURE
PROGRAMRATEMETERPRESCALE SETTINGPROGRAM SCREENSETTING PROCEDURE
When in the Ratemeter Prescale setting screen, when a appears on the rightmost of the display, it is time to set the decimal point position for the prescale
factor. Press thebutton until decimal point is in desired position.
Note: This decimal point is used only for the prescale factor and it will not appear
on the ratemeter screen.

Press thebutton to move the cursor to the next digit.
Press thebutton to set the desired value of a particular digit.
Depressing this button will cause the value to auto scroll.

DECIMAL POINT
POSITION SETTINGRATE DISPLAY
MULTIPLIER(x1 / x10)

In this program mode screen, the Decimal Point Position for the ratemeter runmode display is set.
When in this screen mode, press thebutton to move the decimal point inthe desired location.
In this program mode screen, it is used to select the Rate Display Multiplier.
Selecting x10 will add a zero on the last digit of the display. This zero will notchange value and will not affect the decimal point position.
When in this screen mode, press thebutton to select x1 or x10.
When “1” is displayed, it is x1. When “10” is displayed, it is x10.

CONNECTIONS
Add Input (Voltage Input) G36-101,201,301
8

RST 47INA 36INB 25 Enable/R

GND 1

Supply a Voltage pulse between
terminal l and j (GND) tocount (measure).

Add/Subtract Individual Input (Voltage Input) G36-104
87INA 3

6INB 25 Enable/RSupply a Voltage pulse between
terminal k and j (GND) toAdd, terminal l and j (GND)
to Subtract.

RST 4

GND 190° Quadrature Input (Voltage Input) G36-102
8RST 47INA 3

6INB 25 Enable/R

GND 1

Supply a 90° Quadrature Voltagepulse between terminal l(Signal A) and terminal k(Singal B).

RST 47INA 38RST 47INA 36INB 25 Enable/R

GND 1

Connect or supply an OpenCollector signal between
terminal k and j (GND) tocount (measure).

Add/Subtract Individual Input (Contact/Open Collector Input) G36-103
8RST 47INA 3

6INB 2

5 Enable/R

GND 1

Connect or supply an OpenCollector signal between
terminal k and j (GND) toAdd, terminal l and j (GND)
to Subtract.

Programming Mode (except G36-101)
8

RST 4

7INA 3

6INB 25 Enable/R

GND 1

Connect terminals n and j(GND) to enter into Programming Mode.

Front Reset Enable/Disable (only for G36-101)

Remote Reset (except G36-201)
8Add Input (Contact/Open Collector Input) G36-101,201,301

Connect terminals m and j(GND) to reset the count.

8RST 47INA 36INB 26INB 25 Enable/R

GND 15 Enable/R

GND 1Connect terminals n and j(GND) to enable the Front Resetbutton. Disconnect to disable
the Front Reset button.

CALCULATING THE PRESCALE FACTOR
The Prescale Factor is used to convert the incoming count pulses to the desired unit of measure to be displayed (meters, feet, liters,gallons, etc.) or to correct a known amount of error (wheel wear, viscosity, etc.)<TOTALIZER>
Prescale Setting Range : 0.0001 ~ 99.9999(0.0000 setting will result to a scaling of 100)
Prescale Formula: P S =DPF
PPR

P S = Prescale
D P F = Decimal Point Factor
x x .x x x x 1x x x x x . x 10x x x x . x x 100x x x . x x x 1000x x . x x x x 10000
P P R = Pulse per Revolution(based on sensor Output pulse)
Example: In a conveyor, a sensor produces 20 pulses per 1mmovement. Determine the prescale factor when a 1/10 m
display is required. The resulting PS based on the formula is asfollows:
PS=1020= 0.5000<RATEMETER>
Prescale Setting Range : 0.001 ~ 9999
(0.0000 setting not possible)
Prescale Formula: P S =SECxDPF
PPRP S = Prescale
D P F = Decimal Point Factor
xxxx 1x x x . x 10x x . x x 100x . x x x 1000
S E C = No. of seconds in the rate time unit
(i.e. items/second = 1, items/min =60, items/hour = 3600, etc.)
P P R = Pulse per Revolution(based on sensor Output pulse)
Example: A sensor produces 1 pulse per 1m of material.
Determine the rate prescale factor to display no. of items per
minunte is required. The resulting PS based on the formula isas follows:
PS=60 x 1
1= 60.00

OPERATING PROCEDURETOTALIZER / RATEMETER

G36 - 301

RATEMETERDISPLAYTOTALIZERDISPLAY

Two displays are available, Totalizer and Ratemeter
displays. To toggle between displays, press theT/ Rbutton.
To enter Program Mode, connect Terminals n andj using a jumper wire. After program settings areentered, disconnect the jumper wire to savesettings and display will go back to Count Mode.

BUTTON FUNCTIONSBUTTON

T/ RRST

T/ R
RST

COUNT MODE

PROGRAM MODE
This button is used to move the cursor from one digit to another.
The active digit is indicated by a blinking display.

This button is used to toggle thedisplay between Totalizer andRatemeter displays.

This button is used to set the desired value of each digit.
Keeping this button depressed will cause numbers to autoscroll.
This button is also used to set other program settings like Front Resetenable/disable, Rate Display Multiplier, etc.

This button will reset the the Totalizer.
The ratemeter will not be affected bythis button.

By pressing RST
button repeatedly while keeping the T/ R
buttondepressed, the display will scroll from one program screen to another. There are
6 program screens each indicated by a number on the left most of the display.

NOT APPLICABLE

PROGRAMMING PROCEDURE
PROGRAM(1)
TOTALIZERPRESCALE SETTING(2)TOTALIZERDECIMAL POINT
POSITION SETTING(3)
RATEMETERPRESCALE SETTINGPROGRAM SCREEN
Program No.

Program Item

SETTING PROCEDURE
When the display shows the Prescale Factor setting screen, the right most digit willblink. Choose the digit to set by pressing the T/ R
button. The active digit willblink.
Set the desired value of the chosen digit by pressing the RST
button.
Depressing the RST
button will cause the value to auto scroll.
When the display shows the Decimal Point Position setting screen, press theRST
button until the desired decimal point position is reached.

When in the Ratemeter Prescale setting screen, when a appears on the rightmost of the display, it is time to set the decimal point position for the prescale
factor. Press the RST
button until decimal point is in desired position.
Note: This decimal point is used only for the prescale factor and it will not appear
on the ratemeter screen.

Press the T/ R
button to move the cursor to the next digit.
Press the RST
button to set the desired value of a particular digit.
Depressing this button will cause the value to auto scroll.

(4)
RATEMETERDECIMAL POINT
POSITION SETTING

When the display shows the Decimal Point Position setting screen, press theRST
button until the desired decimal point position is reached.

(5)
RATEMETERRATE DISPLAY
MULTIPLIER(x1 / x10)

When the display shows the Rate Display Multiplier setting screen, press theRST
button to select desired setting.
Selecting x10 will add a zero on the last digit of the display. This zero will notchange value and will not affect the decimal point position.
When “1” is displayed, it is x1. When “10” is displayed, it is x10.

(6)TOTALIZERFRONT RESETSETTING

When the display shows the Front Reset setting screen, choose the desired settingof the Front Reset by pressing the RST
button. A display indicates anenabled front reset. Adisplay indicates a disabled front reset.
Note: The remote reset is still active even if front reset setting is disabled.

DIMENSIONS

3232

40

75

6833

PANEL CUTOUT

67

Specifications and printed contects subject to change without prior notice.
All Rights Reserved, Copyright © 2006 LINE SEIKI CO., LTD. 1ST Edition.
Last Revised March 4, 2006.

LINE SEIKI CO.,LTD.
Address :Contacts :URL:37-7, Chuo-Cho, 2-Chome, Meguro-ku
Tokyo 152-0001 JAPAN
Tel: +81-3-3716-5151 Fax: +81-3-3710-4552
E-mail: webtrade@line.co.jp
http://www.lineseiki.com