LINE SEIKI CO., LTD.

DATA LOGGER SOFTWARE MANUAL

DL2005Address:

Data Logger Software
for Thermometer TC-3200
Instruction Manual

Thank you for purchasing our Data logger software
DL2005 for Thermometer TC-3200.
Please read this instruction manual carefully before
using to ensure the correct usage of this software.
Please keep this instruction manual for futurereference.
Attention- Do not damage the surface of CD-ROM.
- Do not store the CD-ROM where it can be exposed to direct
sunlight, dust, high temperature and high humidity.
- Do not copy or modify this software without our approval.

Specification
Maximum number of acquisition data: 1,600 File format: dat
System Requirement
OS: Windows Vista/7/8/8.1 (Japanese version or English version)
CD-ROM Drive or DVD-ROM Drive
VGA Display 640 x 480 or higher screen resolution
Installation Procedure
1) Run "Setup.exe" in the CD-ROM and follow the instructionsin the installation wizard.

1. Operation Procedures
Before running the software,- Install the driver from CD-ROM supplied with the USB cable
USB-01 into the PC when the USB cable is connected to the PC.
- Insert the thermocouple probes into the sockets of TC-3200.
- Connect TC-3200 to the PC through the USB cable USB-01.
- Choose "PHOTO" on the switch of USB cable.
(1) Preparation
Press [POWER/ESC] Button on TC-3200 for at least 5
seconds until the sub display shows "".
(2) Run the software<In the case it is installed in "Program Files(x86)" folder>1) Open C:/Program files/DL2005/ folder from Start menu.
2) Click the right button on the software.
3) Choose "Run as administrator".
If the property compatibility is changed, double click on thesoftware to run the software.
4) The software screen will appear on the display.
<In the case it is installed in other folder by choosing"Change Directory" when installing.>
1) Open the folder the software was installed.
2) Double click on the software to run the software.
3) The software screen will appear on the display.
(3) Select the COM port
Select the COM port the USB cable USB-01 is connected to.
More detailed information can be found in the manualof USB-01 .
(4) Start acquiring the data
Click "START" button on the software screen.
Click "YES" button when "Data File Check" appearson the display.
(5) Send out data from TC-3200
Press [POWER/ESC] Button on TC-3200 once.
The sub display will show "" and the storage data
will be sent out from the "RS-232 output terminal "(6) View the data
Click "View Data" button on the software screen toview the acquired data.
(7) Finish acquiring the data
Press [FUNCTION] Button on TC-3200 to exit theData Sending function.
(8) Save the acquired data
The acquired data is saved in the "DL2500" folder
automatically. The file name will be YYMMDD.dat.
(YY=Year, MM=Month, DD= Date)
The data will be overwritten on the same file when saving
the data on the same date repeatedly.
Copy the saved file in other folder before acquiring thenew data if you do not want to overwrite.

37-7 Chuo-cho, 2-Chome
Meguro-ku, Tokyo 152-0001 Japan
Tel: +81-3-37165151
Fax: +81-3-37104552
webtrade@line.co.jp
http://www.lineseiki.com

Contact :E-mail :Web:2) Install the software into "Program Files(x86)" folder or other folder you like.
<In the case it was installed in "Program Files(x86)" folder>
It is necessary to click the right button and choose "Run as administrator"
when running the software.
How to change the property compatibility to avoid clicking the right button
every time1) Open "Properties" window by clicking the right button on the softwareand choosing "Properties".
2) Click "Compatibility" tab in the Properties window.
3) Check the option “Run this program as an administrator” in "Privilege
Level" section.
4) Click "OK" to save the change.
<In the case it is installed in other folder by choosing "ChangeDirectory" when installing.>
Run the software in accordance with usual procedure.
Uninstallation Procedure1) Close the software before uninstalling if the software is running.
2) Open "Control Panel" window from Start menu.
3) Open "Programs and Features" or "Uninstall a program" on"Control Panel".
4) Choose "DL2005" and click the "Uninstall" button.

(9) Close the software
Click "Exit" button or the close button

to close the software.

2. Software Screen

aegdhfib
(a)
(b)
(c)
(d)
(e)
(f)
(g)
(h)
(i)
(j)

Comm.Port: Select the COM port.
Start: Start acquiring the data.
View Data: View the data.
Data File Name: The file name that the acquired data is saved
currently
Data Transfer Status: Current status of acquisition process
Number of data: The number of acquired data
Channel: The no. of thermocouple probe in the dual channel
measurement.
Value: Measured temperature
Unit: Measurement unit (Degree C)
Exit: Close the software

3. Importing the dat file into Microsoft Excel
1)
2)
3)
4)
5)
6)
7)

jc

*

Run Microsoft Excel
Choose "File" > "Open" from the menu bar.
Select "All files" in the dropdown list.
Choose the dat file you want to import and click "Open" button.
Text Import Wizard will open.
Choose "Delimited" and click "Next" button.
Choose "Comma" and click "Finish" button.
* Microsoft, Excel and Windows are registered trademarks ofMicrosoft Corporation in the United States and/or other countries.

This manual was last revised Nov. 28, 2014. 4LU029A*Subject to change without prior notice.
All Rights Reserved,Copyright 2014,LINE SEIKI CO.,LTD.