CA9033 C+L Band ASE Broadband
Light Source

Technical Specifications V2.01Mar., 2019www.caoptonicsgroup.com

CA9033 C+L Band ASE Broadband Light Source
The CA OPTRONICS GROUP’s CA9033 C+L band broadband light source is based on
ASE technology. C+L band ASE provides significantly more power density into a singlemode fiber than a regular LED and more than 90 nm wide broadband light source. It offer
superior performance for the test of DWDM/CWDM components, AWG & PLC
components, optical amplifiers, tele-com optical filter and other general purpose of fiber
optical test and measurement applications. It is special design for volume DWDM/CWDM
filter and components production line application.
CA9033 C+L band ASE broadband light source is a High Performance, Small Dimension, Fast
Startup, Affordable Optical Broadband Light Source. It provides 1525 ~ 1610 nm wavelength C+Lband broadband light output.

Features
Wide broadband light output
High spectral density
Quick startup
Small dimension
Affordable priceApplications

DWDM/CWDM filter and components test
Fiber optic parts and components spectrum analyze

Optical Coherence Tomography (OCT)

Fiber Sensing , Fiber Cable test
Fiber Optical , Telcom R & D lab testSpecifications
Model #
Operation Wavelength-10 dB Bandwide
Total Output PowerSpectral Density
Connector Type

CA90331525 ~ 1610 nm>= 85 nm
> 200 mW>= -15 dB/nm
FC/APC (Other Adaptor Can Be Customerized)

Spectral Density Stability within 15< +/- 0.02 dB ***mins
Output Power Stability within 15 mins
< +/- 0.05 dB ***Output Power Stability within 8 hrs<= +/- 0.1 dB ***Operation ModelCW
Fiber TypeSingle Mode SFM-28
Operation Temperature0～ +40℃
Storage Temperature-30～+80℃Power Supply
AC 110 ~ 220 V ±10%, 50Hz, 20W
Dimensions
44 mm H, 224 mm W, 300 mm DWeight
5.0 kg*** Remark: All spectral and power stability test is base on temperature 25 Degree, temperature tolerance +/- 2
degree, power on 40 minutes , CW operation mode condition.

Typical Spectrum:

One CA9033 C+L Real Product Spectrum

CA9034 O Band SLED Broadband Light Source

Technical Specifications V2.01
Mar., 2019www.caoptronicsgroup.com

CA9034 O Band SLED Broadband Light Source
The CA OPTRONICS GROUP’ s CA9034 O band broadband light source is based on
SLED technology. O band SLED provides significantly more power density into a singlemode fiber than a regular LED and more than 90 nm wide broadband light source. It offer
superior performance for the test of DWDM/CWDM components, AWG & PLCcomponents, optical amplifiers, tele-com optical filter and other general purpose of fiber
optical test and measurement applications. It is special design for volume DWDM/CWDM
filter and components production line application.
CA9034 O band SLED broadband light source is a High Performance, Small Dimension, Fast
Startup, Affordable Optical Broadband Light Source. It provides 1260 ~ 1360 nm wavelength O
band broadband light output.

Features
Wide broadband light output
High spectral density
Quick startup
Small dimension
Affordable priceApplications

DWDM/CWDM filter and components test
Fiber optic parts and components spectrum analyze

Optical Coherence Tomography (OCT)

Fiber Sensing, Fiber Cable test
Fiber Optical, Telcom R & D lab testSpecifications
Model #Operation Wavelength-10 dB Bandwide
Total Output PowerSpectral Density
Connector TypeCA90341260 ~ 1360 nm>= 100 nm> 10 mW>= -19 dB/nm
FC/APC (Other Adaptor Can Be Customerized)

Spectral Density Stability within 15< +/- 0.02 dB ***mins
Output Power Stability within 15 mins
< +/- 0.05 dB ***Output Power Stability within 8 hrs<= +/- 0.1 dB ***Operation Model
CW
Fiber Type
Single Mode SFM-28
Operation Temperature
0～ +40℃
Storage Temperature-30～+80℃
Power Supply
AC 110 ~ 220 V ±10%, 50Hz, 20W
Dimensions
44 mm H, 224 mm W, 300 mm DWeight3.0 kg*** Remark: All spectral and power stability test is base on temperature 25 Degree, temperature tolerance +/- 2degree, power on 40 minutes , CW operation mode condition.

Typical Spectrum:

One CA9034 O Band Light Source Real Testing Spectrum

Contact Information
United States:
CA OPTRONICS GROUP INC
3652 Edison Way
Fremont, CA 94536
USA
Tel: 1-510-366-7353
Fax: 1-510-795-1795
www.caoptronicsgroup.com

Product specifications and descriptions in this documentation subject to change without notice.
Copyright @ 2005 CA OPTRONICS GROUP INC.
Nov., 201531000033 V2.01