Accreditation#113037

PRODUCT CATALOG 2024

Automobile

Aircraft

Factory

SAFETY
Life

Railway

Construction

Pedestrian

Motorcycle

every

00

ジャスティ̲プロダクトカタログ̲E.indd 1

2024/02/20 9:53

JASTIPRODUCTCATALOG

INDEX
Corporate Outline

02

JASTI Products

03

Remarkable Features of JASTI s Dummy Products

05

Anthropomorphic Test Device Dummies
Front Impact Test Dummy Hybrid-III

07

THOR 50th Dummy

09

Side Impact Test Dummy

11

FAA Dummy for Aircraft Seat Evaluation

13

Pedestrian Dummy

14

Lightweight Dummy

15

Headform Impactor

16

Pedestrian Legform Impactor

17

Other Products

18

Calibration Equipment

19

Next-Generation Calibration Test Software

21

Skin Damage Test・Sensor

2201

ジャスティ̲プロダクトカタログ̲E.indd 2

2024/02/20 9:53

02

Corporate Identity

03

We cast away "egoism" which allows caring

05

only about ourselves, and keep the mind of"altruism" to sacriﬁce ourselves to help07

others. We consider the feelings of others,

09

11

look around calmly, and extend our hands to13

help others. Every employee in JASTI keeps

14

the mind of "altruism" in his or her hearts,

15

and walks with people through works which

16

can do good in the world.

17

18

19

21

22

Quality PolicyーContribute to society through automobile
collision safety evaluationsー

Action Policies

1. Aim at being one of a kind instead of being the number one
2. Carry out cooperate activities from customers’ viewpoint
3. Increase abilities and strive for continuous quality improvement
4. Maintain fairness and control risks adequately to improvecustomer trust

Goals in Production of Anthropomorphic Dummies
1. Anthropomorphic dummies as measuring apparatus(Reduce individual differences of dummies and control stability of
performance levels)

2. Repeatability(Keep repeatability and reliability of dummies)

3. Anthropomorphic dummies with reproducibility

The history of anthropomorphic dummies.
It is the distinct road that JASTI has taken.

JASTI's Distinct Footsteps
1990

1995

2000

2005

2010

2015
JASTI UTAMA

2020

2023

JASTI USAJASTI Hong Kong

UTAMA Engineering

JASTI China
JASTI Europe

JASTI

02

ジャスティ̲プロダクトカタログ̲E.indd 3

2024/02/20 9:53

JASTI’s Cutting-edge Safety Technologies
the Automobile Industry Worldwide

03

ジャスティ̲プロダクトカタログ̲E.indd 42024/02/20 9:53

Su

es

SupportingHS
HS

Light weight
Hybrid-III
95th Percentile
Male Dummy

ES-2 reSide Impact
Dummy

ES-2
Side Impact
SID-IIs
Dummy
Side Impact
Dummy

THOR 50
Dummy

Light weight
Hybrid-III
5th Percentile
Female Dummy

Hybrid-III 95th
Percentile Large Male
Dummy

Hyblid-III 50th
Percentile Male
Dummy

Hyblid-III 5th
Percentile Female
Dummy

Hybrid-III 3 Year Old
Child Dummy

ジャスティ̲プロダクトカタログ̲E.indd 5

WorldSID 50th
Percentile Male
Dummy

Hybrid-III 5th
Hybrid-III 50th
Male Pedestrian Male Pedestrian
Dummy
Dummy
Hybrid-III 5th
Percentile Female
Pedestrian Dummy
FAA H-III
95th Dummy

THOR-05FDummy

FAA H-III
95th HS Dummy
FAA H-III
50th HS Dummy

Hybrid-III 95th
Percentile Large Male
Pedestrian Dummy

FAA H-III
50th Dummy
Q3 Dummy04

2024/02/20 9:53

Product Features

Remarkable Features of JASTI’s Dummy Products
Safety of Mold Products
Since the company started manufacturing, JASTI has used
DINA and DINP instead of DOP as plasticizers for molded
products, which provide a high degree of safety, however,according to requirements for environmental assessment inEurope (RoHS 2), the cancer-causing property of theseplasticizers is concerned because they contain phthalic acid.
We switched to a new phthalic acid free plasticizer in 2015.
That makes JASTI's PVC products much safer.

Damping Material
JASTI developed a new damping material
which has a performance that is equal toconventional products. The combination ofsteel ribs and this new damping material
enables issuing a Low/High Speed Impact
Certification for displacement of a thorax.

Rib unit test

Features of Rubber Products
Dummies require a high degree of biofidelity and are made up of many kinds of rubber materials to use their properties offlexion, extension, etc.
JASTI offers parts with different hardness for Lumbar Spine, Nodding Block, and Knee Flesh Insert, based on information on
drawings. In a calibration test of a dummy, adjustment and calibration are possible by combining these parts with different
hardness. You are welcome to try and please feel free to contact us.

※These matters are reported at ISO/TC22/SC12/WG5, etc. For details, refer to our website http://www.jasti.co.jp/en.

05

ジャスティ̲プロダクトカタログ̲E.indd 6

2024/02/20 9:53

Product Featuresucts
Different Hardness Types of Rubber Products

Lumbar Spine

Lumbar Spine is based on rubber materials, has a role of the lumbar spine of a human body, and absorbs an impact on thethorax by deforming.
Performance requirements of Lumbar Spine include not only surface hardness, but also repeatability, reproducibility, and durability. Specially, a surface hardness test is required for Hybrid-III 50th Male, and a torso flexion test is required for 5th Female,and Chest Jacket, Rib Assembly, and Lumbar Spine, which are basic components of the dummies, largely relate to thedisplacement of the thorax and limit its properties. Therefore, the precision of each component is obviously important andimproving the precision and reliability of each component secures the reliability of a dummy. For this reason, we always conduct various unique tests to pursuit reliability, durability, and reproducibility.

●Reliable Lumbar Spine●Test on Lumbar Spine●Debond test

Fine adjustment in a neck flexion/extension test

80

82

85

87

90

● Nodding Blocks with diﬀerent hardness for ﬁne adjustment in a neck ﬂexion/extension test

Fine adjustment in a knee impact test
The legal drawing requires a hardness of 40 to 50 (Shore A),
and JASTI offers a Knee Flesh Insert with a hardness of 45,
which is the intermediate value, as the standard product. If arequired impact force cannot be obtained in a performance test,
replace the Knee Flesh Insert with another one with different
hardness for adjustment. JASTI offers 3 different types of Knee
Flesh Insert: Low, Middle, and High. When a required value
cannot be met in a knee impact test, we recommend to use

Low

MiddleHighanother type of Knee Flesh Insert which differs in hardness.

06

ジャスティ̲プロダクトカタログ̲E.indd 7

2024/02/20 9:53

Anthropomorphic Test Device Dummy

Front Impact Test DummyF

Hybrid-III 50th Percentile Male Dummy
Hybrid-III 5th Percentile Female Dummy

H
L

Front Impact Test Dummies, Adult Male/Adult Female,
49CFR PART572 Subpart E/Subpart O, EuroNCAPFrApApplicable standard■日本 TRIAS ■USA FMVSS208 ■EU ECE R94

H

F

Ap

Hybrid-III 50th Percentile Male Dummy

Hybrid-III 5th Percentile Female Dummy

Dummy Specifications
Hybrid-III 50M , 5F Dummy

Outer Size (mm)Sitting Height878 - 889

Shoulder Pivot Height505 - 521

Buttock to Knee Length579 - 605

Knee Pivot Height

485 - 501

Shoulder Width421 - 437

Head Back to Backline

40 - 46

Popliteal Height429 - 455

USA (NHTSA)EuroNCAP

774 - 800

Shoulder Pivot Height

431 - 457

Buttock to Knee Length

520 - 546

Head

Head drop

Same

Neck

Neck Bent/
Extension

Same

Knee Pivot Height393 - 419

ThoraxHigh/(Low Speed)

High /
Low Speed

Shoulder Width

350 - 366

Femur Flexion

Same

Head Back to Backline

43 - 48

Pelvis355 - 376

Friction

Ball Bearing(High / Low)

Popliteal Height

Compression

ImpactKnee Slider
Weight (kg)Outer Size (mm)Sitting Height

Foot

Q

Q

Weight (kg)Head

4.536±0.045

Neck

1.542±0.045

Neck

0.910±0.09

Upper Torso

17.191±0.136

Upper Torso

12.020±0.14

Lower Torso23.042±0.136

Lower Torso

13.250±0.14

Upper Arm (each)

1.996±0.090

Upper Arm (each)

1.180±0.05

Lower Arm and Hand (each)

2.268±0.090

Lower Arm (each)

0.900±0.05
0.280±0.05

Head3.730±0.05

Thigh (each)

5.987±0.090

Hand (each)

Lower Leg and Foot (each)

5.443±0.136

Thigh (each)

3.130±0.09

Total Weight77.700±1.180

Lower Leg

3.270±0.05

Foot0.790±0.05

Total Weight49.050±0.91

07

ジャスティ̲プロダクトカタログ̲E.indd 82024/02/20 9:53

Anthropomorphic Test Device Dummy

Front Impact Test Dummy
Hybrid-III 95th Percentile
Large Male Dummy

Hybrid-III 95th Percentile Large Male Dummy
Outer Size (mm)Sitting Height927 - 942

Shoulder Pivot Height541 - 556

Buttock to Knee Length624 - 650

Knee Pivot Height521 - 546

Front Impact Test Dummy, Adult Large Male

Shoulder Width

467 - 483

Head Back to Backline

86 - 91

Applicable standard ■Designated as SAE standard H3-95 dummy

Popliteal Height457 - 483

Weight (kg)Head

4.94±0.05

Neck

1.68±0.05

Upper Torso

22.32±0.36

Lower Torso

30.30±0.36

Upper Arm (each)

2.81±0.05

Lower Arm and Hand (each)

2.06±0.05

Thigh (each)

8.21±0.09

Lower Leg and Foot

5.75±0.09

Total Weight

101.24±1.63

Hybrid-III 3Year Old Child Dummy
Outer Size (mm)Sitting Height546.1

Shoulder Pivot Height

315.0

Buttock to Knee Length

292.4

Hybrid-III 3 Year Old Child Dummy

Knee Pivot Height249.2

Shoulder Width

244.1

Front Impact Test Dummy, 3 Year-old Child

Head Back to Backline

53.3

Popliteal Height

226.1

Applicable standard■49 CFR PART572 Subpart P

myWeight (kg)
Head

2.72

Neck

0.79

Upper Torso

7.00

Upper Arm (each)

0.44

Lower Arm (each)

0.46

Thigh (each)

1.01

Lower Leg (each)

0.61Foot (each)

0.31

Total Weight

16.17

Q3 Dummy
Outer Size (mm)Q3 Dummy

Q3 Dummy(EU) / Q3s Dummy(USA)

Sitting Height544

Shoulder Pivot Height

329

Chest

142

Shoulder Width

259

Buttock to Knee Length

305

Buttock to Behind the Knee Length

253

Weight (kg)Head + Neck

3.17

Torso (including clothing)

6.40

Upper Arm

0.75

Lower Arm

0.73

Thigh

2.00

Lower Leg

1.54

Total Weight

14.60

08

ジャスティ̲プロダクトカタログ̲E.indd 92024/02/20 9:53

Anthropomorphic Test Device Dummy

Front Impact Test Dummy

T

THOR 50th Percentile Male Dummy
Applicable standard■NHTSA ■EURO NCAPNHTSAEURO NCAP

T

THOR 50th Dummy
Weight (kg)Outer Size (mm)Seated heightL1

Hip pivot heightHip pivot to seat back3.88 - 4.28906±13

Head Assembly

L2

116±5

Neck Assembly

2.66 - 2.94

L3153±13Lower Torso Assembly

14.39 - 15.85

Thigh clearance

L4

183±9

2.69 - 2.97

Knee pivot to bottom of foot

L5

505±7

Lower Arm Assembly(including hands), (left & right)

Knee pivot to hip pivot

L6

413±9

Knee centerline to knee centerline

L7

253±5

Head back to seat backL8

91±1.180

Rib 3 depth

L9Upper Leg Assembly, (left & right)

5.37 - 5.91

Lower Leg Assembly, (left & right)

4.74 - 5.22

Instrument Ground Straps,(4) M6 SHCS, (8) M6 FHCS

0.03 - 0.09

226±9Upper Torso24.85 - 26.25
0.16 - 0.19

Rib 7 depth

L10

229±9

Neck Foam and Skin Assembly

Shoulder-Elbow length

L11

391±7

Front/Rear Panel Assembly, Jacket

1.67 - 1.85
0.21 - 0.27
73.45

Width across arms

L12

461±9

Thermal Pants

Waist width

L13

331±7

Total Weight

Back of elbow to wrist pivot

L14

291±9

Wrist pivot to tip of middle ﬁnger

L15

165±9

NHTSA

EuroNCAP

Spine BoxAll 3 degteeth

Set at9,0,-9,-12deg

Lower Leg

THOR LX
original

Hybrid-III 50th09

ジャスティ̲プロダクトカタログ̲E.indd 102024/02/20 9:53

Anthropomorphic Test Device Dummy

THOR 5th Percentile Female Dummy

A prototype is being developedwith a draft drawing.
Outer Size (mm)Seated height

788

Weight (kg)
Total Weight

48.2

THOR-50Ｍ Finite Element Model
For virtual testing andverification of advanced
safety systems
Main features●Compliant with all calibration conditions for
THOR-50M●Simplified positioning by Positree(TM)●Easy access to standardized sensor output●Verified stability at a higher impact velocity or underimpact conditions with restrained/unrestrained
passengers●Easy to integrate with existing models throughblock numbering

g

h

10

ジャスティ̲プロダクトカタログ̲E.indd 112024/02/20 9:53

Anthropomorphic Test Device DummySide Impact DummyS

ES-2/ES-2re/SID-IIs Side Impact Dummy

W

Ap
■

（E
US NCAP ES-2reEURO NCAP ES-2

Applicable standard ■ECE R95

ES-2 Side Impact Dummy
Outer Size (mm)Applicable standard ■FMVSS214

49CFR PART572 Subpart U

ES-2re Side Impact Dummy
Outer Size (mm)SID-IIs

Applicable standard ■49CFR PART572 Subpart V

SID-IIs Side Impact Dummy
Outer Size (mm)Sitting Height

900 - 918

Sitting Height

900 - 918

Sitting Height772 - 788

Seat to Shoulder Joint

558 - 572

Seat to Shoulder Joint

558 - 572

Shoulder Pivot Heght

437 - 453

Buttock to Front Knee597 - 615

Buttock to Front Knee597 - 615

Buttock to Knee Length

514 - 540

Thorax Width322 - 332

Thorax Width322 - 332

Chest Width851 - 881

Shoulder/Arm Width

461 - 479

Shoulder/Arm Width

461 - 479

Shoulder Width

341 - 357

Abdomen Width273 - 287

Abdomen Width273 - 287

Waist Width

761 - 791

Seat to Sole433 - 451

Seat to Sole433 - 451

Popliteal Height343 - 369

Weight (kg)Weight (kg)Weight (kg)3.70±0.05

Head

4.0±0.20

Head

4.0±0.20

Head

Neck

1.0±0.05

Neck

1.0±0.05

Neck

0.91±0.09

Upper Torso22.4±1.00

Upper Torso22.4±1.00

Upper Torso

11.11±0.20

Arm (each)

1.3±0.10

Arm (each)

1.3±0.10

Lower Torso

12.52±0.18

Abdomen

5.0±0.25

Abdomen

5.0±0.25

Arm (each)

0.91±0.05

Leg (each)

12.7±0.60

Leg (each)

12.7±0.60

Thigh (each)

3.13±0.09

Pelvis

12.0±0.60

Pelvis

12.0±0.60

Lower Leg (each)

3.27±0.09

Total Weight

72.4±1.20

Total Weight

72.4±1.20

Foot (each)

0.79±0.05

Jacket

0.59±0.07

Total Weight44.12±1.09

11

ジャスティ̲プロダクトカタログ̲E.indd 122024/02/20 9:53

WoO

Anthropomorphic Test Device DummySide Impact Test DummyWorldSID 50th Percentile Male Side Impact Dummy W50-00000
Applicable standard■ECE R95（EU）Euro NCAP TB-029 （USA）FMVSS 214

part V

WorldSID 50th Percentile Male Dummy
Weight (kg)Outer Size (mm)4.29±0.05

Sitting Height

869±30

Head

Buttock to Knee Length

670±30Neck

2.86±0.02

Chest/Arm Width468±30Upper Torso

20.56±0.35

Arm Length

330±30Arm

1.77±0.09

Waist Width324±30Lower Torso17.76±0.20

Knee to Sole588±30Thigh (each)

6.71±0.30

Lower Leg (each)

5.09±0.13Suit1.54±0.10

Total Weight73.91±1.02

12

ジャスティ̲プロダクトカタログ̲E.indd 13

2024/02/20 9:53

Anthropomorphic Test Device Dummy

FAA Dummy for Aircraft Seat EvaluationA

FAA H-III 50th Dummy
FAA H-III 95th Dummy

P

D

Dummy for Aircraft Seat Evaluation

Pe
by*The FAA Dummy is standardized in Part 23.562, 25.562, 27.562, and 29.562 of CFR 14.
*HS (Heli safe) speciﬁcation models are available both for 50th and 95th FAA Dummies.

H

FAA H-III 50th Dummy

FAA H-III 95th Dummy
Outer Size (mm)Outer Size (mm)927〜942

Sitting Height

878〜889

Sitting Height

Shoulder Pivot Height

505〜521

Shoulder Pivot Height541〜556

Buttock to Knee Length579〜605

Buttock to Knee Length624〜650Knee Pivot Height

485〜501

Knee Pivot Height521〜546

Shoulder Width421〜437

Shoulder Width467〜483

Head Back to Backline

40〜46

Head Back to Backline

86〜91

Popliteal Height429〜455

Popliteal Height457〜483

Weight (kg)Weight (kg)Head

4.536±0.045

Head4.94±0.05

Neck

1.542±0.045

Neck

1.68±0.05

Upper Torso

17.191±0.136

Upper Torso

22.32±0.36

Lower Torso23.042±0.136

Lower Torso

30.30±0.36

Upper Arm (each)

1.996±0.090

Upper Arm (each)

2.81±0.05

Lower Arm and Hand (each)

2.268±0.090

Lower Arm and Hand (each)

2.06±0.05

Thigh (each)

5.987±0.090

Thigh (each)

8.21±0.09

Lower Leg and Foot5.443±0.136

Lower Leg and Foot

Total Weight77.700±1.180

Total Weight

5.75±0.09
101.24±1.63

H
FAA H-III 50th HS Dummy

FAA H-III 95th HS Dummy

Outer Size (mm)HS

Outer Size (mm)927〜942

Sitting Height

878〜889

Sitting Height

Shoulder Pivot Height

505〜521

Shoulder Pivot Height541〜556

Buttock to Knee Length579〜605

Buttock to Knee Length624〜650Knee Pivot Height

485〜501

Knee Pivot Height521〜546

Shoulder Width421〜437

Shoulder Width467〜483

Head Back to Backline

40〜46

Head Back to Backline

86〜91

Popliteal Height429〜455

Popliteal Height457〜483

Weight (kg)Weight (kg)Head

4.536±0.045

Head4.94±0.05

Neck

1.542±0.045

Neck

1.68±0.05

Upper Torso

17.191±0.136

Upper Torso

22.32±0.36

Lower Torso23.042±0.136

Lower Torso

30.30±0.36

Upper Arm (each)

1.996±0.090

Upper Arm (each)

2.81±0.05

Lower Arm and Hand (each)

2.268±0.090

Lower Arm and Hand (each)

2.06±0.05

Thigh (each)

5.987±0.090

Thigh (each)

8.21±0.09

Lower Leg and Foot5.443±0.136

Lower Leg and Foot

Total Weight77.700±1.180

Total Weight

5.75±0.09
101.24±1.63

13

ジャスティ̲プロダクトカタログ̲E.indd 14

2024/02/20 9:53

Anthropomorphic Test Device Dummy

Advanced Models of Dummies
Pedestrian Dummy

Dummies for Pedestrian Protection Performance Test
Pedestrian dummies are standing posture dummies which are modiﬁed versions of Hybrid-III 5F, 50M, and 95LM dummies, createdby replacing some parts in the Pelvis, Lumbar Spine, and Knee Slider.

Hybrid-III 5th Percentile Female Pedestrian Dummy
Outer Size (mm)Hybrid-III 5th Male Pedestrian Dummy
Outer Size (mm)Sitting Height774〜800

Sitting Height

Shoulder Pivot Height

431〜457

Shoulder Pivot Height774〜800
431〜457

Buttock to Knee Length

520〜546

Buttock to Knee Length

520〜546

Knee Pivot Height

393〜419

Knee Pivot Height

393〜419

Shoulder Width350〜366

Shoulder Width350〜366

Head Back to Backline

43〜48

Head Back to Backline

43〜48

Popliteal Height355〜376

Popliteal Height355〜376

Head3.730±0.05

Head

Neck

0.910±0.09

Neck0.910±0.09

Upper Torso

12.020±0.14

Upper Torso

12.020±0.14

Lower Torso

13.250±0.14

Lower Torso

13.250±0.14

Upper Arm (each)

1.180±0.05

Upper Arm (each)

1.180±0.05

Lower Arm (each)

0.900±0.05

Lower Arm (each)

0.900±0.05

Hand (each)

0.280±0.05

Hand (each)

0.280±0.05

Thigh (each)

3.130±0.09

Thigh (each)

3.130±0.09

Lower Leg

3.270±0.05

Lower Leg

3.270±0.05

Foot0.790±0.05

Foot0.790±0.05

Total Weight49.050±0.91

Total Weight49.050±0.91

Weight (kg)Weight (kg)3.730±0.05

Hybrid-III 95th Percentile Large Male Pedestrian Dummy

Hybrid-III 50th Male Pedestrian Dummy
Outer Size (mm)Outer Size (mm)927〜942

Sitting Height

878〜889

Sitting Height

Shoulder Pivot Height

505〜521

Shoulder Pivot Height541〜556

Buttock to Knee Length579〜605

Buttock to Knee Length624〜650Knee Pivot Height

485〜501

Knee Pivot Height521〜546

Shoulder Width421〜437

Shoulder Width467〜483

Head Back to Backline

40〜46

Head Back to Backline

86〜91

Popliteal Height429〜455

Popliteal Height457〜483

Weight (kg)Weight (kg)Head

4.536±0.045

Head4.94±0.05

Neck

1.542±0.045

Neck

1.68±0.05

Upper Torso

17.191±0.136

Upper Torso

22.32±0.36

Lower Torso23.042±0.136

Lower Torso

30.30±0.36

Upper Arm (each)

1.996±0.090

Upper Arm (each)

2.81±0.05

Lower Arm and Hand (each)

2.268±0.090

Lower Arm and Hand (each)

2.06±0.05

Thigh (each)

5.987±0.090

Thigh (each)

8.21±0.09

Lower Leg and Foot5.443±0.136

Lower Leg and Foot

Total Weight77.700±1.180

Total Weight

5.75±0.09
101.24±1.63

14

ジャスティ̲プロダクトカタログ̲E.indd 15

2024/02/20 9:53

Anthropomorphic Test Device Dummy

H

Lightweight Dummy

Th

Light weight Hybrid-III 5th Percentile Female Dummy（20Kg）
Light weight Hybrid-III 95th Percentile Male Dummy（40Kg）

JA

pe

ap

im

Th

【Application】

oth

This dummy is lightweight, easy-to-handle, and suitable for in-vehicle comfort studies and the like.

I

The range of motion of each part is the same as that of an anthropomorphic test dummy.

Light weight Hybrid-III
5th Percentile Female Dummy（20Kg）■ Total weight : 20kg

Light weight Hybrid-III
95th Percentile Male Dummy（40Kg）■ Total weight : 40kg

The head can be moved 60°to the left and right.

15

ジャスティ̲プロダクトカタログ̲E.indd 16

2024/02/20 9:54

Headform Impactor

The pedestrian headform impactors are dummies replicating heads of human bodies used for "pedestrian protection
performance tests" required by car assessment programs such as NCAP in each country. They were developed by
JASTI in cooperation with JARI and JAMA, became the new pedestrian protection safety standard, and were later
approved by ISO. They are ejected to make a collision mainly with the hood or the windshield of a car to measure theimpact value in G and HIC (Head Injury Criteria). That enables measuring a head injury value in a pedestrian accident.
They are also used for safety evaluations of playground equipment including mats in places such as playgrounds,
other than automobile safety evaluations.

ISO Type*EC type products which are in compliance with
European laws and regulations are also available.

ISO（EC）Adult 4.5 kg

4.5 kg Comp Adult
ISO-A-CS

4.5 kg Adult Skull BIA Type
ISO-A-C-BIA

4.5 kg Skin
ISO-A-S

4.5 kg Skull
ISO-A-C

ISO（EC）Child 3.5 kg

3.5 kg Child Comp
ISO-C-CS

3.5 kg Child Skull BIA Type
ISO-C-C-BIA

3.5 kg Child Skin
ISO-C-S

3.5 kg Child Skull
ISO-C-C

4.5 kg,3.5 kg Skin
ISO-AC-S(Adult Child common)

FMVSS201

FMVSS226

Free Motion Headform Impactor

Mitigation Headform Impactor

16

ジャスティ̲プロダクトカタログ̲E.indd 17

2024/02/20 9:54

Pedestrian Legform Impactor

O

Flex PLI - GTR
Since it is highly possible that a pedestrian receives a severe injury to his or her legs in a vehicle-pedestrian collision accident,
UNECE WP29/GRSP released standards for protecting pedestrians' legs in 2002 and a flexible type of legform impactor wasadded to the Global Technical Regulations (GTR) in 2009 according to the proposal of JAMA and JARI. The reason of the
proposal is that it is more biofidelic than a rigid impactor and enables measuring leg injuries appropriately.
Since 2013, after specifications of Flexible Pedestrian Legform Impactor (Flex PLI-GTR), which was evaluated by organizations
including Flex TEG, were determined, JASTI has been engaged in developing and manufacturing the final specification model.
At the end of 2015, we completed evaluation of prototypes both in Japan and overseas and started manufacturing and selling
the products in 2016. They are currently used in Japan, China, India, and Europe.

The knee joint part is composed of machined aluminum knee

Flex PLI-GTR

block of which top and bottom ends bonded with 12 spring wires,and 4 displacement gauges measure the amount of elongation
displacement of the knee joint.

Outer Size (without flesh) (mm)
Leg Length

982.0

the femur and tibia and the amount of displacement of the knee,

Femur Length to Knee Joint

433.0

its structure allows attaching a data collection device inside theTibia Length to Knee Joint

495.0

Knee Width

118.0

Leg Width

84.0

Knee Depth

108.0

Leg Depth

90.0

Femur Length from Top of Knee Assembly

339.0

Tibia Length from Bottom of Knee Assembly

404.0

To facilitate ease of data collection for the bending moment ofknee and ensures a high degree of flexibility in handling.

Knee Block

Femur & Bone Core

Calibration Test on Flex PLI-GTR

Tibia

Knee

We have made many improvements in the product to pass
the calibration test. Please see how the test was performed.

Femur

Pendulum17

ジャスティ̲プロダクトカタログ̲E.indd 18

2024/02/20 9:54

Other Product

H-III 50th Ankle Bumper
For Non Metal Contact(J2949)A conventional ankle bumper is attached to an ankle lower shell, but the contact between metal parts of the ball ankle shaft andthe lower shell caused noises. To prevent the noise, we offer an ankle bumper integrated with an ankle lower shell. This integrated ankle bumper is the standard specification in IIHS in the United States.

H-III 50M/5F/95LM
Zippered Lower Leg Flesh

H-III 50M/5F/95LM Velcro
Joined Lower Leg Flesh

Other Products

Ball Baring Knee Slider

Chairs for safekeepingwith casters and adjusters

H-III 50M

H-III 5F

dummy

dummy

Euro NCAP
According to the standard specifications of Hybrid-III 50th
Male Dummy and 5th Female Dummy, a friction knee
slider is installed in both of them, however, with the newly
developed ball bearing knee sliders, both 50M and 5F canaddress needs in the market.

18

ジャスティ̲プロダクトカタログ̲E.indd 19

2024/02/20 9:54

Calibration And Certification

Head Drop Test System

Neck Flex/
Extension Test System

Thorax Impact Test System●Test objects
Front drop test：Hybrid-Ⅲ 5F/50th/95th、World SID50th、THOR50th
Side drop test：ES-2、ES-2re、SID-Ⅱs●Test objects
Hybrid-Ⅲ 5F/50th /95 th、ES-2、World SID50th、THOR50th

Front impact test dummySide impact test dummy

Knee Impact / Share Test

Hip Joint Test System

Torso Flexion Test

●Test objects
Hybrid-Ⅲ 5F/50th

You can see a movie of the testfrom the QR code on the right.

You can see a movie of the testfrom the QR code on the right.

19

ジャスティ̲プロダクトカタログ̲E.indd 20

2024/02/20 9:54

mCalibration And Certification

Rib Drop Test Equipment

Foot Impact Test

Foot Compression Test

●Test objects

ECE R94

CFR PART572
Subpart E

ES-2、ES-2re20

ジャスティ̲プロダクトカタログ̲E.indd 21

2024/02/20 9:54

Next-Generation Calibration Test SoftwareS

Next-generation calibration test softwarethat uses a web server

Features1. Introduction of a web server enables users to view and print out test data and reports using
a browser from anywhere without installing any special software.
2. As functions to control the test, import test data, as well as create, view, and print reports
are separated, test systems can be configured more flexibly.
3. Introduction of a full-scale database enables large-capacity, safer test data management.

Support for consistency in calibration tests
JASTI conducts calibration tests on all dummies and parts that require a calibration test certification based on CFR 49 Part 572 before delivery.
However, there are many cases in the market where the same result as a calibration test certification cannot be reproduced. Many sites have theproblem of difference in reproducibility even though they meet the calibration test standard in CFR 49. To minimize the problem, it is necessary toachieve consistency among many test conditions such as calibration test equipment, data acquisition system, test software, test environment, andskills. As a dummy manufacturer, we have a vast amount of experience not only with consistency in dummies, but in achieving consistency incalibration test equipment, to minimize the problem (difference). Please feel free to consult us for any problem no matter how small. As a dummy
manufacturer, JASTI will support customers for consistency in test conditions.
■ Please contact us for details of the specifications.

21

ジャスティ̲プロダクトカタログ̲E.indd 222024/02/20 9:54

Skin Damage Test・Sensor

Skin damage test
For testing if skin gets torn when any bodypart gets caught

Evaluation equipment A(for adults)

Evaluation equipment A(for children)

Sensor、Transducer、Potentiometer■ES-2(re) Side Impact DummyLinear Position Transducer■SID-IIs Side Impact Dummy
1/2" Potentiometer■H-III 50M・5F・95LM Dummy
Chest Rotary Potentiometer

ery.

he
to

■Knee Potentiometer & String■SPIRAL Track®

nd
in

my

ns.

22

ジャスティ̲プロダクトカタログ̲E.indd 23

2024/02/20 9:54

Jasti Co., Ltd.

jasti.co.jp/en
New company building set to open in 2024

Head Office

Technical Center

2-4-3 Miyoshi, Koto-ku, Tokyo 135-0022, Japan
TEL: +81-(0)3-5245-3661 / FAX: +81-(0)3-5245-8596

669 Yanagisawa, Numazu, Shizuoka 410-0308, Japan
TEL: +81-(0)55-966-8620 / FAX: +81-(0)55-966-8629

JASTI CO.,LTD
You can make it with JASTIJAST will cooperateI with your every
demand for the human safety.
Printed Feb.2024-Ver.EN／100

ジャスティ̲プロダクトカタログ̲E.indd 24

2024/02/20 9:54