enElcometer 213/2
Digital Waterproof Thermometer

Operating Instructions

enRDeclaration of Conformity:
This product complies with the requirements of the following EU Directives:
2014/30/EU EMC Directive
2011/65/EU Low Voltage Directive
2011/65/EU Restriction of the Use of certain Hazardous Substances in Electrical and Electronic Equipment (RoHS) - Directive
A CE Declaration of Conformity is available to download via:
https://downloads.elcometer.com/Declaration_of_Conformity/English/DoC_213_2.pdf
This product complies with the requirements of the following UK Standards
S.I. 2016 No. 1091
Electromagnetic Compatibility Regulations 2016
S.I. 2016 No. 1101
Electrical Equipment (Safety) Regulations 2016
S.I. 2012 No. 3032
Restriction of the Use of Certain Hazardous Substances in Electrical and Electronic Equipment (RoHS)
Regulations 2012
A UKCA Declaration of Conformity is available to download via:
https://downloads.elcometer.com/Declaration_of_Conformity/English/DoC_213_2_UKCA.pdf
This product is Class B, Group 1 ISM equipment according to CISPR 11.
Class B product: Suitable for use in domestic establishments and in establishments directly connected to a low voltage power supply network
which supplies buildings used for domestic purposes.
Group 1 ISM product: A product in which there is intentionally generated and/or used conductively coupled radio frequency energy which isnecessary for the internal functioning of the equipment itself.

Ris a registered trademark of Elcometer Limited. All other trademarks acknowledged.

© Copyright Elcometer Limited. 2022
All rights reserved. No part of this Document may be reproduced, transmitted, transcribed, stored (in a retrieval system or otherwise) ortranslated into any language, in any form or by any means (electronic, mechanical, magnetic, optical, manual or otherwise) without the priorwritten permission of Elcometer Limited.

Doc No. TMA-0757 Issue 01
Text with Cover No. 32121

Section
1234567
89PageAbout your gauge . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2
Fitting a battery . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3Fitting a probe . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4Taking a reading . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5
Changing measurement units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Error messages . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Maintenance/Storage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7
Spare parts and accessories . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Technical specifications. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 91en

RCONTENTS

en

R

Thank you for your purchase of this Elcometer 213/2 Digital Thermometer. Welcome to Elcometer.
Elcometer are world leaders in the design, manufacture and supply of inspection equipment for coatingsand concrete. Our products cover all aspects of coating inspection, from development, through application
to post application inspection.
Your Elcometer 213/2 Digital Thermometer is a world beating product. With the purchase of this product you
now have access to the worldwide service and support network of Elcometer. For more information visit ourwebsite at www.elcometer.com

1 ABOUT YOUR GAUGEThe Elcometer 213/2 Digital Thermometer is a simple, easy-to-use, battery powered digital thermometer
which allows quick and easy measurements of surface, air and liquid temperature as well as thetemperature of soft materials.
1.1 GAUGE FEATURES•

Waterproof case offering IP66/67 protection•Range -49°C to +1372°C (the maximum temperature is dependant on probe type)•Extruded aluminium case for superior durability•

Rubber bumper seals for impact resistance•Easy to read LCD display•High accuracy2•

Elcometer 213/2 Digital Thermometer withbatteries fitted•

Operating instructions

Note: A probe is not supplied with thethermometer. Probes must be ordered separately see “Spare parts and accessories” on page 8.
The Elcometer 213/2 Digital Thermometer is packed
in a cardboard and foam package. Please ensurethat this packaging is disposed of in anenvironmentallysensitivemanner.
Pleaseconsult
yourlocal Environmental Authority forfurther guidance.

2 FITTING A BATTERY
When the battery voltage becomes low, ‘LO BAT’ is shown on the display. The instrument will continue
to measure accurately while ‘LO BAT’ is shown, but changing the battery at the earliest
convenience is recommended.
When the battery voltage becomes too low for the gauge to measure accurately, 'Flat' and then 'Bat' isshown on the display and the gauge will then switch off. The battery must be replaced at this time tocontinue using the gauge.
3en

R

1.2 WHAT THE BOX CONTAINSenR

To change the battery:
1. Using a Posidrive (crosshead) screwdriver, remove the two battery compartment cover retaining
screws on the base of the gauge and release the battery pack.
2. Fit the new batteries into the battery clips taking care to observe the correct polarity.
3. Replace the battery compartment cover ensuring that the seals are not damaged and are correctly
positioned.
4. Tighten the retaining screws firmly to ensure that the seals are compressed and will provide a waterresistant seal.
Note: Batteries must be disposed of carefully to avoid environmental contamination. Please consult yourlocal Environmental Authority for information on disposal in your region.
Do not dispose of any batteries in fire.

3 FITTING A PROBEYour Elcometer 213/2 Digital Thermometer should only be used with a 'K Type' thermocouple probe madeto BS EN 60584:1996 which is fitted with a suitable miniature thermocouple plug. A range of suitable probes
is available from your Elcometer supplier or direct from Elcometer - see “Spare parts and accessories” onpage 8.
Probes are fitted to the socket at the top of the gauge. When fitting a probe, never force the probe connector
into the socket if the connector and socket do not easily join. Make sure the connector matches the socketand that it is properly aligned and undamaged before fitting.
44 TAKING A READING
1. To switch on the gauge, press
2. Apply the tip of the probe to the substance, medium or surface to be measured.
The reading on the display may take a number of seconds to stabilise depending on the nature of themeasurement and the sensitivity of the probe.
3. If required, to freeze the reading on the display, press
4. Press. ‘HOLD’ is shown on the display.

again to return the gauge to normal measurement.

5. To switch offa the gauge, pressa.

The gauge will switch off automatically after 20 minutes of inactivity.

5en

R

en

R

5 CHANGING MEASUREMENT UNITS
To change the instrument units from °C to °F or vice versa:
1. Switch off the gauge.
2. Pressbut do not release the button.

3. While continuing to press
4. Continue to press, press

to switch on the gauge.

for seven seconds until the units change in the display.

5. Release

6 ERROR MESSAGESThe following error messages may be shown on the display:‘LO’:

You are attempting to measure below the specified temperature range of the gauge‘HI’:

You are attempting to measure above the specified temperature range of the gauge‘Err’:

No probe is fitted or there is a fault with the probe

6You own one of the finest digital thermometers in the world. If looked after, it will last a lifetime.
•Be careful not to scratch the LCD screen or break the glass screen.

•Do not subject the gauge to a strong impact.

•Clean the gauge with a soft, dry cloth such as those made to clean eyeglasses. Do not use solvents!
This gauge incorporates a Liquid Crystal Display. If the display is heated above 50°C (120°F) it may
be damaged. This can happen if the gauge is left in a car parked in strong sunlight.

Regular calibration checks over the life of the gauge are a requirement of quality management procedures,e.g. ISO 9000, and other similar standards. For checks and certification contact Elcometer or yourElcometer supplier. Your gauge does not contain any user-serviceable components .In the unlikely eventof a fault, the gauge should be returned to your Elcometer supplier or directly to Elcometer. The warrantywill be invalidated if the instrument has been opened
Contact details can be found on the outside cover of these operating instructions and atwww.elcometer.com

7en

R

7 MAINTENANCE/STORAGE

en

R

8 SPARE PARTS AND ACCESSORIESThe following spare parts and accessories are available direct from Elcometer or your local supplier:

8Beschrijving

Reeks

Artikelnummer

Magnetic Surface Probe, 13mm
(0.51”) Diameter-50°C to 150°C (-58°F to 302°F)

T99911728

Surface Probe, 130 x 4.2mm(5.11 x 0.17”) Diameter-50°C to 600°C (-58°F to 1112°F)

T2136069-Liquid Probe,130 x 3mm (5.11 x0.12”) Diameter-200°C to 1100°C (-328°F to 2012°F)

T9996390-

Needle Probe, 130 x 3mm (5.11 x0.12”) Diameter-50°C to 400°C (-58°F to 725°F)

T2136391-

Measurement Range:-49°C to +1372°C (-56°F to +2500°F)

Resolution:

0.1°C (0.1°F) up to +299.9°C (+ 599.9°F)
1°C (1°F) above +299.9°C (+ 599.9°F)

Accuracy:±1% of the reading ±1 digit

Probe:‘K type’ Thermocouple

Ambient Temperature:

0°C to +50°C (0°F to +122°F)

Case Dimensions:

35 mm x 60 mm x 115 mm (1.4" x 2.4" x 4.5")

Weight:194 g (6.9 oz)

Display:

12 mm LCD

Battery:

3 x AAA (LR03) 1.5V

Battery Life:

Approximately 5000 hours

9en

R

9 TECHNICAL SPECIFICATIONSen

R10