EM-1000

DIGITAL MULTI-FUNCTION ENVIRONMENT METER MANUAL
EM-1000
Digital Multi-Function Environment Meter

ATTENTION

Please note that misuse of this device may lead toinjury to the user or damage to the device.
Please observe all safety precautions and warnings
in this instruction manual.

Customer Service

Thank you for purchasing our Digital Multi-Function
Environment Meter.
Please read this instruction manual carefully before using
to ensure the correct usage of this device.
Please keep this instruction manual for future reference.

For safe usage of this device, please observe all statements regarding
precautions and warnings in this instruction manual.

ATTENTION

Operation- Do not use this device near machines that emit strong
electromagnetic fields or objects that store static electricity.
- Do not drop or subject this device to strong impact.
- Do not use or store this device where it will be exposed towater or in places with wet conditions.
- Do not use or store this device where it can be exposed todirect sunlight, dust, high temperature and high humidity.
- See the battery case markings to ensure that the battery isproperly installed.
- For accurate measurements, do not subject this device inmeasuring sudden change of temperatures.
- Remove the battery when the device will not be used for along period of time.
- Do not attempt to disassemble or modify this device.

Contact
E-mail :Web:1.) Insert the thermocouple plug into the input terminal.
2.) Press the "ON" button to turn on the meter.
3.) Select the Temperature function by pressing " Function Button ".
4.) Engage the thermocouple probe to the object to be measured.
5.) After the display stabilize, take the reading of the temperature.
6.) Press the "°C/°F Button " to select °C/°F unit.

e. Light Measurement
1.) Press the "ON" button to turn on the meter.
2.) Select the Light meter function by pressing " Function Button ".
3.) Press the " Lux/Ft-cd Button "to select measuring unit " Lux " or" Ft-cd ".
4.) Face the "Light Sensor" to the light source. Get reading whendisplay stabilizes.
The light display digits are oriented 180° from the other function
displays for easy exposure and output reading of the light sensor.

a. Humidity & Ambient Temperature Measurement
1.) Press the "ON" button to turn on the meter.
2.) Select the Relative Humidity function by pressing " Function
Button ".
3.) The LCD display will show the current measurement indicatedby the unit of measurement symbol "%RH" and "°C".
(The upper display shows the Humidity measurement, lower
display shows Temperature measurement.)
4.) Press the "°C/°F Button " to select °C/°F unit.

11c. Air Flow (CFM. CMM) Measurement▲

1.) Press the "ON" button to turn on the meter.
2.) Select the Air Flow function by pressing " Function Button ".
3.) Press the " CFM/CMM Button" to select unit.
4.) Press the " Area-Set Button" to select area size.
5.) Press each button to set the desired area size.
Move the cursor to the next digit▲Increase the selected digit.
▼Decrease the selected digit.
6.) Press the “Enter Button” to save the value.
7.) Press the “ESC button” to return to normal mode.
If the “ESC button” is pressed before pressing “Enter Button”,
the display returns to normal mode without saving the value.
8.) Face the " Anemometer Sensor " to the measuring airflow.
Get reading when display stabilizes.
For the CMM measurement, the area size is " meter square ",the lower display will show " m-2 ".
The area setting size is from 0.001 to 30.000 meter square.
For the CFM measurement, the area size is " feet square ",the lower display will show " F-2 ".
The area setting size is from 0.01 to 322.92 feet square.

ModelEM-1000

Display

8 mm LCD display

Measurement

Anemometer, Humidity, Temperature,

12345Range

Accuracy

Power Consumption

Approx. DC 6.2 mA

32 - 122 °F± 2.5 °FWeight160g ( battery included )0 - 50 °C± 1.2 °C

Dimension ( HWD )156 x 60 x 33 mm ( 6.14 x 2.36 x 1.29 inch )Measurement

Range

AccuracyLight

0 - 2,0000 Lux± 5% rdg ± 8 dgt

b. Resolution ( 23 ± 5°C )MPH

0.9 - 67.0 MPH

0.1 MPH

9knots

0.8 - 58.3 knots

0.1 knots

12Temperature

32 - 122°F

2( thermister )Measurement

70% RH : ±4% RH
70% RH :0.1 km/h±( 4% rdg + 1.2 % RH)

Humidity

0 - 1,860 Ft-cd
Measurement

Range

Accuracy

Temperature-148 - +2,372 °F± ( 1% rdg + 2°F )( Type K )-100 - +1,300 °C± ( 1% rdg + 2°C )0.1 °F

Measurement

Area setting

0 - 50°C

0.1 °C

CMM

0.001 - 30.000 meter square

Range

Resolution

CFM

0.01 - 322.92 feet square

54,000 CMM

0.001 - 1 CMM

1,908,400 CFM

0.001 - 100 CFM

Range

Resolution% RH

10 - 95 % RH

0.1 % RH

Temperature

32 - 122 °F

0.1 °F( thermister )0 - 50 °C

0.1 °C

Range

Resolution

0 - 2,200 Lux

1 Lux

1,800 - 2,0000 Lux

10 Lux

0 - 204.0 Ft-cd

0.1 Ft-cd* auto range

170 - 1,860 Ft-cd

1 Ft- cd

Measurement

Range

Resolution

Temperature-148 - +2,372 °F

0.1 °F( Type K )-100 - +1,300 °C

0.1 °C

Measurement
CMM
cube meter / mincube feet / min

Back View

1.) To start recording the measurement data, press the "REC button" once.
The display will show the "REC" symbol on the display.
2.) While the "REC" symbol is on display, press the "REC button" once todisplay the Maximum measured data recorded. The "MAX" symbol willalso show on the display. Press the "REC button" once again to display
the Minimum measured data recorded. The "MIN" symbol will also showon the display. To view back "Max"("Min") measured data, press "RECbutton",alternately. To clear the stored MAX/MIN values, while on the"MAX" display (if Maximum value is to be cleared) or on the "MIN"
display (if Minimum values is to be cleared), press continuously the"HOLD button" until value is cleared and the display goes back to"REC" display.
3.) To exit from Data Record function, press the "REC button" for at least2 seconds and the display will go back to current measuring display.

± 1.2 °C

006P DC 9V battery ( Heavy duty type )1.4 - 108.0 km/h

4. Data Record (Max / Min Reading)

0 - 50°C

Power Supplykm/h1.) While measuring, press the "HOLD button" to hold the last measured
value on the display. The display will also show the "HOLD" symbol.
2.) Press the "HOLD button" again to exit from the data hold function.

± 2.5 °F

10 - 95 % RH

0.1 m/s

3. Hold Function

32 - 122°F

Indication of "- - - - "0.4 - 30.0 m/s

Function/ ▼ Button
Anemometer Sensor
Thermocouple Input Socket
Humidity Sensor
Light Sensor
LCD display
Battery Compartment / CoverHand Strap

0.9 - 67.0 MPH

Over Input Display

Measurementm/s

67891011121320 m/s : ±4% F.S.

0 - 50°C ( 32 - 122°F )Air flow

ON/ESC Button
Hold Button
REC/Enter Button
Unit/Zero/( CFM/CMM ) Button.
°C/°F/ ▲Lux/Ft-cd
Area-set Button

0.4 - 30.0 m/s

Operating Temperature

1.0 ft/min

Front View

20 m/s : ±3% F.S.

Max. 80% RH.

CFM1380 - 5,910 ft/min

Operating Humidity

80 - 5,910 ft/min

10Accuracy

0.8 - 58.3 knots

Light, Air flow ( CFM / CMM ).

5b. Air Velocity & Ambient Temperature Measurement

Air velocity

ft/min3614Range

1.4 - 108.0 km/h

a. General Specifications

Air velocity

8Measurement

6. Specifications

Resolution

7EM-1000c. Accuracy

This unit will automatically switch off if none of the buttons are pressed
for approximately 10 minutes.
Note: The Auto Power Off feature is disabled during Data Recordfunction.

Range

2. Parts

1. Operation Procedures

1.) Press the "ON" button to turn on the meter.
2.) Select the Anemometer function by pressing " Function Button ".
3.) Press the " Unit/Zero Button "to select unit of Air Velocity.
4.) Press the "°C/°F Button " to select °C/°F unit.
(The upper display shows the Air Velocity measurement,
lower display shows Temperature measurement.)5.) Face the " Anemometer Sensor " to the measuring airflow.
Get reading when display stabilizes.

37-7 Chuo-cho, 2-ChomeMeguro-ku, Tokyo 152-0001 Japan
Tel: +81-3-37165151
Fax: +81-3-37104552
webtrade@line.co.jp
http://www.lineseiki.com

5. Auto Power Off

d. Temperature Measurement (K type Thermocouple probe)▲

Safety Precautions

Address:LINE SEIKI CO., LTD.

DIGITAL MULTI-FUNCTION ENVIRONMENT METER MANUAL

Measurement

Humidity

Measurement
Lux
Light
Ft-cd

Remark :ft/min : feet per minutem/s : meters per second
km/h : kilometers per hour

MPH : miles per hourknots : nautical miles per hour
Ft-cd : feet candle

7. Battery Replacement
1.) When the LCD display shows “LO BAT” , the battery needs to be
replaced.
2.) Open the screw of the battery cover by screwdriver, then remove
the battery.
3.) Replace the battery with a new one and position this new battery
correctly inside the meter’s battery compartment.
4.) Fasten back the cover.
The instrument cannot measure accurately after low battery indicatoris displayed. Please replace the battery immediately.

8. Storage
For proper storage, avoid places where the device can be exposed todirect sunlight, high humidity, high temperature, vibration and shock,dust, rust, corrosion, etc. Remove battery when the device will not be
used for a long period of time.

9. Accessories
Battery x 1
Hand Strap x 1
Instrcution manual x 1* K type Thermocouple probe sold separetely

This manual was last revised Mar. 21, 2013. 4LU017A*Subject to change without prior notice.
All Rights Reserved,Copyright © 2013,LINE SEIKI CO.,LTD.