TempStick Probe with handle

TempStick Probe with handle temperature data logger is aminiaturized temperature recorder with 2 meters cable andrigid metal probe of 28 cm length, with a wide range ofmonitoring applications especially in the food, environment
and pharmaceutical industry, for continuous monitoring in
fridges and fridge cells according to ISO 9001 and 12830
standards, HACCP, FDA, BRC ecc. certification. Its reduced
size and no need of external power supply make theTempStick system extremely versatile. The data logger canbe supplied with calibration certificate with Accredia(Italian reference center, NIST equivalent) traceability.
Available also in the following versions:
TempStick: with internal sensor
TempStick Probe: with cable, rigid probe
TempStick Probe with thin cable: with thin cable, rigidprobe
TempStick Probe IP68: with cable, rigid probe, protcted in an IP68 box
TempStick Probe -80: with cable, rigid probe, for temperatures from -80°C
TempStick Probe 200: with cable, rigid probe, for temperatures up yo 200°C

Applications

Cold Chain

Food & Beverages●●Environment●●●Laboratories●

Main features
●●
●
●
●

●

Mission programming and reading using a PC through the TecnoStick Interface or SRI and FLI portableinterfaces
Reduced size for easy use for every need
Useful handle for the rigid probe
Quick response timeStand-by function: start the logger and then, using the metallic clip, set it to a stand by status. In this way
you can send many pre-programmed loggers to the sender of the products without providing him with thesoftware and the interface. Before sending the goods he will just have to remove the clip to start theacquisitions
Works with StickLog Pro and FridgeLog Z software

Plus●
●
●
●
●
●
●

High accuracy and precision
Product's core temperature controlled directly
Easy to use with SRI and FLI portable interfaces
Compliant with HACCP and EN12830 regulations
MKT formula implemented, for quick data analysis, also on SRI and FLI
Provided with calibration certificate Accredia (NIST equivalent) traceable (on demand)
Long life battery

The system
The basic system, for transport, is made up by:
TempStick datalogger
StickLog Pro software
TecnoStick Interface or SRI●●●Warehouses

Pharmaceutical

The FridgeLog Z system, for continuous monitoring, is made up by:
TempStick datalogger
FridgeLog Z sotware
FLI interface●●●Accessories
●
●
●
●
●
●

StickLog ProTecnoStick Interface
SRI
FridgeLog Z
FLI
TempStick metal clip

Technical specifications
Dimensions

50 X 24 X 10 (mm)Probe dimensions

Cable 2000 l X 3 Ø (mm) - Probe 85 l X 4 Ø (mm)Weight65 gr

Temperature range-30 °C ÷ +60 °C

Sensor temperature range-40 °C ÷ +90 °C

Standard calibration points (temperature)-30/0/20/60°C

Extra calibration points (temperature)

Within the range -40 °C ÷ +90 °C

Temperature resolution

0,03 °C

Temperature accuracy± 1 °C without certificate / ± 0,25 °C with Accredia traceable certificate (valid in thecalibration range)

Memory (n. of acquisitions)

2.730

Acquisition step

From 1 every minute to 1 every 255 minutes

Battery life

Up to 10 years or 3 millions of acquisitions

Protection degree

IP63

Accessories

TecnoStick Interface, SRI, FLI