S-Micro

S-Micro is a temperature data logger from 0°C to 220°C(standard calibration from 25°C to 140°C, calibrable from0°C to 220°C, only the probe will resist above 140°C) with
20 or 50 or 100 or 150 mm or on demand length external
probe (probes cannot be switched) on a 5 mm base (baseheight is not counted for probe length), managed with
Windows software and USB interface. With every S-Microa battery kit for battery replacement is included.
Battery is user replaceable and the data logger isprovided with an Accredia (NIST equivalent) traceablecetificate on 6 points.

Applications

Food & Beverages

Sterilisation

The version for up to 100°C only is available too.
It is part of a series of data loggers divided in P-Micro (up to100°C) and S-Micro (up to 140°C): there are different
versions with standard length probes or on demand.
P-Micro and S-Micro versions are the smallest loggers forsuch temperatures available on the market. L and XL
versions feature a longer life battery and their operative
range is extended (can be calibrated from -40°C; standard
calibration range is 25 °C ÷ +100 °C for P-Micro and 25 °C
÷ +140 °C for S-Micro). They require an interface for PC
connection: DiskInterface HS Mini (for P-Micro and S-Microonly, not for L and XL versions), DiskInterface HS andMultibay.

Pasteurisation

MedicalPharmaceutical

There are also other models of high temperature data loggers, for pressure and humidity too.
Validation

Main features
●
●
●
●
●
●
●
●
●

●

Extremely reduced size: the smallest volume available for high temeprature loggers
With different lengths rigid probe for penetration
Completely food grade and waterproof
All software calculate lethality value (F0, PU, A0 ecc.)
Low battery consumption for an extended battery life
User replaceable battery (software shows battery status)
Very easy to deploy in any type of package
Accredia (NIST equivalent) traceable calibration certificate included
Available extended calibration from 0°C to 220°C (order extra calibration points; in case of wide
calibration range the accuracy might be worse)
Heat protections available for use above 140°C

Plus
●

●
●
●

Extremely high accuracy and precision: with an accuracy of ± 0,1°C these devices can be employed in any
application involving food, pharmaceuticals, validation, laboratory and medical field
Fast response time thanks to the 3 mm diameter probe
Printed reports compliant with health regulations and ISO (data are not editable in the software)
NO NEED TO DRILL ANY LIDS, NO NEED TO MODIFY YOUR PRODUCT PACKAGING: PROGRAM THE DATALOGGER, PLACE IT YOU PRODUCT AND START YOUR PROCESS!

Healthcare

Laboratories

The system
The system is made up by:
S-Micro temperature data logger
DiskInterface HS Mini or DiskInterface HS or Universal Multibay
SPD software or TS Manager software (compatible with the FDA 21 CFR Part 11 regulation)●●●Accessories
●
●
●
●
●
●
●
●
●
●
●
●
●

SPDTS Manager
DiskInterface HS Mini
DiskInterface HS
Universal multibay
Fixing base
Adhesive rubber for high temperatures
Locking bolt
Fastening system
Teflon protective tube
S-Micro heat protection 1
S-Micro heat protection 2
Battery kit for P-Micro, S-MicroTechnical specifications
Dimensions

14 h X 17 Ø (mm)Probe dimensions

Probe base dimensions 5 h X 14 Ø (mm) - Probe 20/50/100/150 l X 3 Ø (mm) - Probe l on
demand X 3 Ø (mm) (l on demand: min. 10 mm / max. 175 mm. For longer probes ask forquotation)

Weight

13 grMaterials

Stainless steel AISI316L, PEEK

Temperature range

0 °C ÷ +140 °C

Sensor temperature range

0 °C ÷ +220 °C

Standard calibration points (temperature)

25/50/75/100/125/140°C

Extra calibration points (temperature)

Within the range 0 °C ÷ +220 °C

Extra calibration points (temperature)

Up to 220°C (using the heat protections)Temperature resolution

0,01 °C

Temperature accuracy± 0,1 °C (within the calibration range)

Memory (n. of acquisitions)20.224

Acquisition step

From 1 every second up, with 1 second steps

Protection degree

IP68

Battery life

560.000 acquisitions at 1 second step continuously (calculated time @25°C. Battery life
may be shorter if used in low temperatures)

Software&Mobile App

SPD, TS Manager

Accessories

DiskInterface HS Mini, DiskInterface HS, Universal Multibay