1. Check contents and Symbol Key.
(See Overview)
2. Fully charge Type 4448 unit(s).
(See Charge)
3. Turn on the unit(s).
(See turn On and Off)4. Configure unit(s) to desired use
parameters.
(See Configure at Startup)
5. Calibrate unit(s).
(See Calibrate)
6. Mount units on the subject.
Optional: Change mounting clips.
(See Mount)
7. Perform the measurements.
(See Measurement Control)
8. Access and review your data.
(See Review Measurement Data andDownload Data to a PC)

Copyright © Brüel & Kjær. All rights reserved.

NOTE: Consult Type 4448 User Manual for safety information
1

EnglishFirst UseCharge











Symbol key

EnglishOverview

To Charge Type 4448

To Link Chargers

Place the unit on the provided charger.
Full charging from flat will takeapproximately 90 minutes.
Charging: Flashing red light.
Charged: Constant blue light.
Ensure that Type 4448 has enoughcharge for the required measurement
duration by looking at the screen,
which will show the run time available
for the current charge.

Link the chargers using cable AO-1493
then connect power supply ZG-0864 toa charger with a free AC input (any AC
input is fine).
The power supply can drive a chain of
up to four chargers.

 Left Key Right Key
 Status Indicator Display Windshield Infrared Sensor

 Press once Press and holdPress multiple times(example shows three times)

2

3

Configuration at Startup
To Turn On Type 4448

Press the Left (L) key. Type 4448 will run througha start-up sequence.
The first screen displays model number andfirmware version. The second screen shows user
defined text (can be configured using Type 7825
Protector).
After startup, the instrument enters stand-bymode, alternatingly showing time and date and(if available) data from the last run.

To Turn Off Type 4448
Hold down the ‘L’ key for 3 seconds.

To Enter Configuration
Mode
Press and hold the ‘R’
key during startup.

Parameters


3 sec

4

To Change ValueCONTRASTLANGUAGE
PRO. MODEDISPLAY MODE
dB ALARMS
PRESET MODE
AUTO
CLEAR MEMORY

Press the ‘L’ key tochange the parametervalue.

Englishturn On and Off

To Select ParameterPress the ‘R’ key toconfirm the value
and move to the next
parameter.
Confirming the final
screen returns the unit
to standard operation.

NOTE: If the memory is to be cleared, a second
screen appears for confirmation. Change the option
to ‘Yes’ by pressing the ‘L’ key, then press ‘R’ toconfirm deletion.

5

Windshield

EnglishCalibration

To Remove the Windshield
Twist the black plastic ring(not the foam) counterclockwise and lift it off.

NOTE: By default Type 4448
is set up for a calibration levelof 94dB. You can change thisusing PC Software Type 7825
Protector (for more details see
Type 7825 Online Help andType 4448 User Manual).

To Attach the Windshield

To Calibrate Type 4448
1. Switch on Type 4448 and wait for it
to start up.
2. Remove windshield(Important: see Windshield)
3. Place the acoustic calibrator
over the microphone as shown
to the left. Be careful to push
the calibrator straight onto Type4448 without twisting, since thismight loosen or over tighten the
microphone capsule.

6

4. Turn on the calibrator. Type4448 automatically recognises
calibration tones and enterscalibration mode.
5. Press the ‘R’ key to confirm thatyou wish to calibrate. Type 4448
will take a few seconds to automatically adjust to the required level.
Once calibration is completed, a ‘CAL
OK’ message will appear. Type 4448
stores the calibration date, time andlevel then returns to the default
screens.

Place the tabs into the slots
then gently twist the black
plastic ring (not the foam)clockwise.

NOTE: When attaching or removing the windshield always hold and twist using theblack plastic ring at the base of the windshield: Never use the foam.

7

To Mount the Units
1. Select the correct clips for theapplication (more or less intense
physical activity, environment andsubject’s clothing requirements). For
example, pins provide a stronger bondbut penetrate protective materials.
2. Place the instrument on the shoulder
ensuring that the microphone points
upwards and that the microphone tip
is about 10 cm from the ear.
For more mounting information,
see manual.

To Remove or Switch Mounting Clips
1. Locate and remove the screws(see image) and lift off the clips.
2. Place the tabs of the new clips into theholes, gently push the clips into placeand insert and tighten the screws.

Measurement Control
To Start a Measurement


3 sec

To Lock and Unlock Keys andDisplay during Measurement


3 sec

Press and hold both keys for 3
seconds; a countdown will be
displayed on screen.
During a measurement, ► will be
displayed in the top left of the screen.
The screen will then cycle between
the dB level, measurement durationand the memory/battery status.

To Stop a Measurement
Press and hold both keys for 3
seconds; a countdown will be
displayed on screen.

8

EnglishMount
Hold the ‘R’ key down and press the‘L’ key 3 times. A padlock symbol willappear. The keys are now locked, anddB values will be hidden.
To unlock the keys, hold the ‘R’ key
down and press the ‘L’ key 3 times.

Pre-set Measurement Time
When using preset measurement
time, the unit can enter locked mode
(optional) during measurement. The
measurement will stop at the settime.
To stop the measurement in advance,
unlock Type 4448 as described above.

9

Download Data to PC

When Type 4448 is not measuring, thescreen will cycle through status informationand data from the previous measurement
run. The data displayed will depend on theconfiguration of Type 4448, either ISO or
OSHA data. The data shown is displayed to
the left.

ISO
MEMORY

10Hrs
80Hrs

OSHA
MEMORY 13:45:03

 13:45:03

DURATION
07:45:12

DURATION
07:45:12

LAEQ 89.9
LCPK 101.4  dB

LAVG 111.4
LZPK 119.4  dB

PA2Hrs

3.20

OSHA DOSE
4.3  %PROJ DOSE
352.5  %PROJ DOSE
352.5  %

LAF 112.3  dB

10To Freeze the Display

10Hrs
80Hrs

LAS 109.3  dB

EnglishReview Measurement Data

To Download Data to a PC
1. Ensure the infra-red download
adaptor (AO-1492) is connected to
the USB port of your PC and Type4448 is sufficiently charged.
2. Start Type 7825 Protector softwareon the PC and Type 4448.
3. Hold Type 4448’s infra-red port nearto and facing the PC’s infra-red
adaptor.
4. Start download sequence within
Type 7825 Protector and follow
instructions for downloading data.

Error Messages
If any error messages occur during
the procedures described in this field
guide, please refer to Troubleshootingin the Type 4448 User Manual.
A flashing red LED indicates either low
battery or memory.

Press the ‘R’ key on any of these screens andthe display will freeze for 10 seconds andthen resume.

11

Field Specifications
BATTERY LIFE
28 hours

TIME HISTORY / MEMORY CAPACITY
180 hours with 1 minute logging period

PRESET MEASUREMENT TIME
00:00–12:00 in 30-minute steps

MEASURING RANGE
Linear Operating Range:
Peak Measurement Range:

OPERATING TEMPERATURE
0 to +40°C

HUMIDITY
30% – 95% RH (non-condensing)

12

65.0 – 140.3dB (RMS)
95.0 – 143.3dB (Peak)1. Inhalt kontrollieren und Symbolerklärung lesen.
(Siehe Übersicht)
2. Typ 4448-Gerät(e) voll aufladen.
(Siehe Laden)
3. Geräte(e) einschalten.
(Siehe Ein- und Ausschalten)
4. Die gewünschten Parameter einstellen.
(Siehe Konfiguration beim Start)
5. Gerät(e) kalibrieren.
(Siehe Kalibrierung)
6. Gerät an der Person befestigen.
Optional: Befestigungsclip wechseln.
(Siehe Befestigung)7. Messungen durchführen.
(Siehe Mess-Steuerung)
8. Daten betrachten und übertragen.
(Siehe Messdaten betrachten und
Datenübertragung zum PC)

HINWEIS: Sicherheitshinweise siehe Bedienungshandbuch für Typ 4448.
13

DeutschErster Einsatz

Laden











Symbolerklärung

 Linke Taste Rechte Taste Statusanzeige
 Display Windschirm Infrarotsensor

Typ 4448 aufladen

Ladegeräte verbinden

Das Gerät auf das mitgelieferte
Ladegerät setzen. Volles Aufladen aus
dem entladenen Zustand dauert ca. 90
Minuten.

Die Ladegeräte mit Kabel AO-1493
verbinden und das Netzteil ZG-0864 an
ein Ladegerät mit freiem AC-Eingang
anschließen (jeder AC-Eingang ist
geeignet).
Das Netzteil kann bis zu vierLadegeräte in Reihe versorgen.

Beim Laden: Rotes Blinklicht.
Voll geladen: Blaues Dauerlicht.
Dafür sorgen, dass Typ 4448 für die
gewünschte Messdauer ausreichendgeladen ist. Auf dem Display wird die
für den derzeitigen
Ladezustand verfügbare
Messzeit angezeigt.

 Einmal drücken

 Drücken und halten
Mehrmals drücken
(im Beispiel dreimal)

14

15

DeutschÜbersicht

Typ 4448 einschaltenDie linke Taste (L) drücken. Typ 4448 durchläuft
eine Startsequenz.
Die erste Anzeige zeigt Modellnummer undFirmwareversion. Die zweite Anzeige zeigt
benutzerdefinierten Text (kann mit Typ 7825
Protector konfiguriert werden).
Nach der Anlaufphase geht das Gerät in den
Standby-Modus, in dem abwechselnd Uhrzeit/
Datum und (sofern vorhanden) Daten derletzten Messung angezeigt werden.

Typ 4448 ausschalten
Die ‘L’-Taste 3 Sekunden gedrückt halten.

Konfiguration beim Start
In den Konfigurationsmodus gehenBeim Start die ‘R’-Taste
drücken und halten.

Parameter


3 sec

16

Wert ändern

KONTRASTSPRACHEPRO. MODUSANZEIGEMODUS
dB ALARMEVOREINSTELLUNGAUTOSPEICHER LÖSCHEN‘L’-Taste drücken,
um den Wert desParameters zu ändern.

Parameter wählen

DeutschEin- und Ausschalten

‘R’-Taste drücken, um
den Wert zu bestätigen
und zum nächsten
Parameter zu gehen.
Bei Bestätigung kehrt
das Gerät mit derletzten Anzeige in den
Standardbetrieb zurück.

HINWEIS: Wenn der Speicher gelöscht werden soll,
ist dies auf einer zweiten Anzeige zu bestätigen.
Mit der ‘L’-Taste ‘Ja’ wählen und dann die ‘R’-Taste
drücken, um das Löschen zu bestätigen.

17

Kalibrierung

WindschirmWindschirm abnehmenWindschirm aufsetzen

Typ 4448 kalibrieren
1. Typ 4448 einschalten und die
Anlaufphase abwarten.
2. Windschirm entfernen(Wichtig: siehe Windschirm)
3. Den akustischen Kalibrator auf das
Mikrofon setzen (siehe links). Beim
Aufsetzen auf Typ 4448 darf der
Kalibrator nicht gedreht werden,
da andernfalls die Mikrofonkapsel
gelockert oder zu fest angezogen
werden könnte.

18

DeutschDen schwarzen Kunststoffring(nicht den Schaumstoff)
entgegen dem Uhrzeigersinn
drehen und abnehmen.

HINWEIS: Standardmäßig ist
Typ 4448 auf einen Kalibrierpegel
von 94dB eingestellt. Dies kannmit der PC-Software Typ 7825
Protector geändert werden(weitere Einzelheiten siehe
Online-Hilfe von Typ 7825 undBedienungs-handbuch von
Typ 4448).

4. Den Kalibrator einschalten.
Typ 4448 erkennt automatischKalibriertöne und geht in den
Kalibriermodus.
5. ‘R’-Taste drücken, um die
Kalibrierung zu bestätigen. Nachwenigen Sekunden stellt sichTyp 4448 automatisch auf den
gewünschten Pegel ein.
Nach der Kalibrierung erscheint die
Meldung ‘KAL OK’. Der 4448 speichert
Datum/Uhrzeit/Pegel der Kalibrierung
und kehrt zur Standardanzeige zurück.

Die Vorsprünge in die
Aussparungen einpassen undden schwarzen Kunststoffring(nicht den Schaumstoff) im
Uhrzeigersinn drehen.

HINWEIS: Beim Aufsetzen oder Abnehmen des Windschirms immer den schwarzen
Kunststoffring unten am Windschirm festhalten und drehen: Nicht am Schaumstoffteil
festhalten oder drehen.
19Befestigung

Mess-Steuerung

Geräte befestigen

Eine Messung starten

Montageclips entfernen oderaustauschen
1. Schrauben entfernen (siehe Abb.)
und Clips abnehmen.
2. Die Laschen der neuen Clips in die
Löcher stecken, die Clips vorsichtig
einschieben, die Schrauben einstecken
und festziehen.

20


3 sec



Deutsch

1. Die geeigneten Clips für die
Anwendung wählen (mehr oderweniger intensive physische Aktivität,
Umgebungsbedingungen und
Anforderungen an die Kleidung).
Beispielsweise ergeben Anstecknadelneine festere Verbindung, doch sie
durchdringen Schutzmaterial.
2. Das Gerät so auf der Schulteranbringen, dass das Mikrofon nachoben zeigt und seine Spitze sich ca. 10
cm vom Ohr befindet.
Weitere Informationen zur Befestigung
siehe Bedienungsanleitung.

Tasten und Display während der
Messung verriegeln



3 sec

Beide Tasten 3 Sekunden drücken undhalten. Auf dem Display erscheint ein
Countdown.
Während der Messung ist oben links
auf dem Display ein ► zu sehen.
Die Anzeige wechselt zwischen dem
Pegel in dB, der Messdauer und dem
Speicherstatus/Batteriezustand.

‘R’-Taste gedrückt halten und 3-mal
die ‘L’-Taste drücken. Es erscheint ein
Vorhängeschloss-Symbol. Die Tasten
sind verriegelt und die dB-Werte nicht
zu sehen.
Zum Entriegeln die ‘R’-Taste gedrückt
halten und 3-mal die ‘L’-Taste
drücken.

Eine Messung stoppen

Vorgewählte Messzeit

Beide Tasten 3 Sekunden drücken undhalten. Auf dem Display erscheint ein
Countdown.

Bei vorgewählter Messzeit lässt sichdas Gerät bei der Messung verriegeln(optional). Die Messung stoppt, wenn
die eingestellte Zeit verstrichen ist.
Um die Messung vorher zu stoppen,
ist Typ 4448 zu entriegeln, siehe oben.
21Datenübertragung zum PC

Wenn Typ 4448 nicht misst, wechselt die
Anzeige zwischen Statusinformation
und Daten der vorherigen Messung. Die
angezeigten Daten hängen von der Konfiguration des 4448 ab (entweder ISO- oder
OSHA-Daten). Welche Daten angezeigt
werden, ist links dargestellt.

ISO

OSHA

10Hrs
SPEICHER 80Hrs

10Hrs
SPEICHER 80Hrs

 13:45:03

 13:45:03

MESSZEIT
07:45:12

MESSZEIT
07:45:12

LAEQ 89.9
LCPK 101.4  dB

LAVG 111.4
LZPK 119.4  dB

PA2Hrs

3.20

OSHA DOSE
4.3  %PROJ DOSE
352.5  %PROJ DOSE
352.5  %

LAF 112.3  dB

22

LAS 109.3  dB

Anzeige anhaltenDaten zum PC übertragen
1. Dafür sorgen, dass der InfrarotAdapter (AO-1492) mit dem
USB-Anschluss des PC verbunden
und Typ 4448 ausreichend geladen
ist.
2. PC-Software Typ 7825 Protector
starten und Typ 4448 einschalten.
3. Typ 4448 so halten, dass sein
Infrarotport aus geringer
Entfernung auf den InfrarotAdapter des PC zeigt.
4. Downloadsequenz in Typ 7825
Protector starten und der Anleitung
für die Datenübertragung folgen.

DeutschMessdaten betrachten

FehlermeldungenDie ‘R’-Taste auf einer der Anzeigen drücken.
Dann bleibt die Anzeige 10 Sekunden stehen,bevor sie wieder zu wechseln beginnt.

Sollten bei den in dieser Kurzanleitung
beschriebenen Vorgängen Fehlermeldungen erscheinen, sehen Sie bitte
unter Fehlersuche im Bedienungshandbuch von Typ 4448 nach.
Eine blinkende rote LED bedeutet
entweder schwache Batterie oderfehlenden Speicherplatz.

23

Technische Kurzdaten
Batterielebensdauer
28 Stunden

ZEITVERLAUF / SPEICHERKAPAZITÄT
180 Stunden mit 1 Minute Abspeicherintervall

MESSZEITVORWAHL
00:00–12:00 in 30-Minuten-Schritten

MESSBEREICH
Linearer Arbeitsbereich:
65,0 – 140,3dB (Effektivwert)
Bereich der Spitzenschallpegel: 95,0 – 143,3dB (Spitzenwert)

BETRIEBSTEMPERATUR
0 bis +40°C

FEUCHTE
30% – 95% RH (ohne Kondensation)24

1. Revise contenidos y Símbolos de las Teclas.
(Ver Información General)
2. Cargue completamente la(s) unidad(es)Dosímetro 4448.
(Ver Carga)
3. Encienda la(s) unidad(es).
(Ver Encendido y Apagado)
4. Configure la(s) unidad(es) con losparámetros deseados.
(Ver Configuración Inicial)
5. Compruebe la(s) unidad(es) “Sólo Laboratorios de Calibración/ Verificación”.
(Ver Calibrado)
6. Monte las unidades en el sujeto.
Opcional: Cambie las sujeciones demontaje. (Ver Montaje)
7. Realice las medidas.
(Ver Control de Medida)
8. Acceda a sus datos y revíselos.
(Ver Revisar Datos de Medida yDescarga de Datos a PC)

NOTA: Consulta el manual del Dosímetro 4448 para ver la información de seguridad.
25

Espanol

Antes de usar su dosímetro por primera vez

Carga











Símbolos de las Teclas

 Tecla Izda
 Tecla Drcha Indicador de Estado Pantalla
 Pantalla Antiviento Sensor de Infrarojos

Para cargar el Dosímetro 4448

Para acoplar cargadores

Coloque la unidad en el cargador. Con
la unidad totalmente descargada, la
carga completa tardará aproximadamente 90 minutos.
Cargando: Luz roja parpadeante.
Cargado: Luz azul constante.
Asegúrese de que el Dosímetro 4448
tiene suficiente carga para la duración
requerida de la medida mirando en lapantalla, que mostraráel tiempo de funcionamiento disponible
de la unidad.

Conecte los cargadores usando el
cable AO-1493. Después conecte lafuente de alimentación ZG-0864 a un
cargador con una entrada de AC libre(cualquier entrada de AC valdrá).
La fuente de alimentación puede
alimentar hasta cuatro cargadores en
cadena.

 Pulse una vez

 Presione y mantenga pulsado
Presione varias veces(el ejemplo muestra tres)26

27

Espanol

Información General

Para Encender el Dosímetro 4448

Presione la Tecla Izquierda (Izda). El Dosímetro
4488 comenzará la secuencia de encendido.
La primera pantalla muestra el número delmodelo y la versión del firmware. La segunda
pantalla muestra un texto definido por el
usuario (puede ser configurado usando elPrograma 7825 Protector).
Después de encenderse, el instrumento pasa
a modo stand-by, mostrando (si es posible),alternativamente la fecha y la hora de la última
medida.

Para apagar el Dosímetro 4448
Presione y mantenga pulsada la Tecla ‘Izda’durante 3 segundos.


3 sec

28

Configuración Inicial
Para Entrar en el
Modo Configuración

Para Cambiar un ValorPara seleccionar un
Parámetro
Espanol

Encendido y Apagado

Presione y mantega
pulsada la Tecla ‘Drcha’durante el encendido.

Parámetros
CONTRASTEIDIOMA
PRO. MODE
VER MODO (ISO/ OSHA)SCROLL LCD
dB ALARMA
TIEMPO FIJOAUTOBORRAR MEMO

Pulse la Tecla ‘Izda’ paracambiar el valor de unparámetro.

Pulse la Tecla ‘Drcha’para confirmar el valor
y moverse al siguienteparámetro.
Confirmando la pantalla
final, la unidad vuelve
al modo estándar defuncionamiento.

NOTA: Si la memoria debe ser vaciada, una segunda
pantalla aparecerá para confirmarlo. Cambie la
opción a ‘Yes’ o ‘Si’ presionando la Tecla ‘Izda’.
Después presione ‘Drcha’ para confirmar el borrado.
29

Calibrado_ “Sólo Laboratorios de Calibración/ Verificación”

Pantalla Antiviento

NOTA: Por defecto el Dosímetro
4448 está preparado para un nivel decalibración de 94dB. Puede cambiar
este utilizando el Programa para PC
7825 Protector (para más detalles, ver
Ayuda Online y Manual de Usuario delPrograma 7825 Protector). Si quiere
realizar una comprobación, encienda el
dosímetro y póngalo a medir, coloque elcalibrador y compruebe que el nivel esde 94 ±0.5. (usuarios)

Para quitar la Pantalla
Antiviento

Espanol

Gire el anillo de plástico negro(no la espuma) en sentido
antihorario, y retire la pantalla.

Para poner la Pantalla
Antiviento

Para Calibrar el Dosímetro 4448
1. Presione el botón de encendido delDosímetro 4448 y espere a que secomplete la secuencia.
2. Quite la Pantalla Antiviento(Importante: ver Pantalla
Antiviento)
3. Coloque el calibrador acústicoacoplándole el micrófono como
se muestra en la imagen. Tenga
cuidado de empujar el calibrador
fuerte, pero sin girarlo, ya que
podría dañar gravemente la cápsuladel micrófono.
4. Encienda el calibrador. El Dosímetro
4448 reconocerá automáticamente
30

los tonos de calibración y entrará
en el Modo de Calibración.
5. Pulse la Tecla ‘Drcha’ para
confirmar que desea calibrar.
El Dosímetro 4448 tardará
unos segundo en ajustar
automáticamente el nivel
requerido.
Una vez que haya sido completada lacalibración, un mensaje de ‘CAL OK’aparecerá. El Dosímetro 4448 almacena
la fecha de calibración, hora y fecha, ynivel. Después, vuelve a la pantalla por
defecto.

Coloque las guías en las
ranuras. Después, suavemente,
gire el anillo de plástico negro(no la espuma) en sentido
horario

NOTA: Cuando esté poniendo o quitando la Pantalla Antiviento, sujete siempre y gire elanillo de plástico negro, en la base de la Pantalla, nunca la espuma.

31Para montar las unidades
1. Seleccione las sujeciones correctas parael uso (actividad física más o menos
intensa, requerimientos del ambientey de la ropa del sujeto) Por ejemplo,
los pins proporcionan un fijado
robusto, pero penetran los materiales
el protección.
2. Coloque el instrumento en el hombro
asegurándose de que el micrófono
apunta hacia arriba y de que su punta
está situada a aproximadamente a 10cm del oído.
Para más información de Montaje, ver elmanual.

Para Quitar o Cambiar las Sujeciones
de Montaje.
1. Localice y quite los tornillos (verimagen) y suelte los clips.
2. Coloque las guías de los nuevas clips
en los agujeros, empuje suavemente
los clips a su sitio e inserte y apriete lostornillos.

32

Control de Medida
Para Comenzar una Medida


3 sec


3 sec

Para Bloquear y Desbloquear Teclas y
Pantalla durante la Medida

EspanolMontajePresione y mantenga ambas teclas
durante 3 segundos; una cuenta
atrás aparecerá en pantalla.
Cuando se esté llevando acabo una medida, el símbolo► aparecerá en la esquina
superior izquierda de la pantalla.
La pantalla mostrará consecutivamente el nivel de dB, laduración de la medida y el estado
de la memoria y de la batería.

Para Parar una Medida
Presione y mantenga ambas teclas
durante 3 segundos; una cuenta
atrás aparecerá en pantalla.

Presione y mantenga la Tecla ‘Drcha’ y pulse
la Tecla ‘Izda’ 3 veces. Aparecerá el símbolo
de un candado. Las Teclas están ahora
bloqueadas, y los valores de dB ocultos
Para desbloquear las Teclas, presione y
mantenga la Tecla ‘Drcha’ y pulse la Tecla‘Izda’ 3 veces.

Preseleccionar Tiempo de Medida
Cuando use un Tiempo de Medida
Preseleccionado, podrá entrar en el modo
bloqueado (opcional) durante la medida.
La medida se detendrá transcurrido eltiempo fijado.
Para detener la medida en curso,
desbloquee el Dosímetro 4448 como se
describe anteriormente.

33

Descarga de Datos a PC

Cuando el Dosímetro 4448 no está midiendo,
la pantalla mostrará consecutivamente
información de estado y datos de la anterior
medida. Los datos mostrados dependeránde la configuración del Dosímetro 4448,
datos ISO o bien datos OSHA. Estos datos semuestran a la izquierda de la pantalla.

ISO

OSHA

10Hrs
MEMORIA 80Hrs

10Hrs
MEMORIA 80Hrs

 13:45:03

 13:45:03

DURACION
07:45:12

DURACION
07:45:12

LAEQ 89.9
LCPK 101.4  dB

LAVG 111.4
LZPK 119.4  dB

PA2Hrs

3.20

OSHA DOSE
4.3  %PROJ DOSE
352.5  %PROJ DOSE
352.5  %

LAF 112.3  dB

34

LAS 109.3  dB

Para congelar la Pantalla

Para Descargar los Datos a un PC
1. Asegúrese de que el adaptador
de infrarrojos (AO-1492) estáconectado al puerto USB del PCy de que el Dosímetro 4448 tiene
suficiente batería.
2. Inicie el Programa 7825 Protector
en el PC y encienda el Dosímetro
4448.
3. Mantenga el puerto de infrarrojosdel Dosímetro 4448 cerca yenfrentado al puerto de infrarrojosdel PC.
4. Comience la secuencia de descarga
con el Programa 7825 Protector ysiga las instrucciones de descargade datos.

Espanol

Revisar Datos de Medida

Mensajes de Error

Pulse la Tecla ‘Drcha’ en cualquiera de estas
pantallas y ésta se congelará durante 10
segundos y después continuará con lo queestaba mostrando.

Si ocurriera algún error durante losprocedimientos descritos, en esta
guía de campo, por favor, consulte lasección de Troubleshooting o solución
de problemas en el Manual de Usuariodel Dosímetro 4448.
Un LED rojo parpadeante indica
que la batería o la memoria estánagotándose.
35Especificaciones
DURACIÓN DE LA BATERÍA
28 horas

HISTORIA TEMPORAL / CAPACIDAD DE MEMORIA
180 horas con 1 minuto de tiempo de registro

TIEMPO DE MEDIDA PRESELECCIONADO
00:00–12:00 En pasos de 30 minutos

RANGO DE MEDIDA
Rango de Linealidad:		
Rango de Medida de Pico:65.0 – 140.3dB (RMS)
95.0 – 143.3dB (Pico)

TEMPERATURA DE FUNCIONAMIENTO
0 a +40°C

HUMEDAD
30% ­– 95% RH (sin condensación)361. Vérifier le contenu et les boutons.
(Voir Vue synoptique)
2. Recharger l’appareil.
(Voir Recharger la batterie)
3. Mettre l’appareil en marche.
(Voir Commutateur Marche/Arrêt)4. Paramétrer l’appareil.
(Voir Configurer au démarrage)
5. Calibrer l’appareil.
(Voir Calibrage)
6. Fixer l’appareil sur la personne.
Facultatif : Changer les pinces defixation. (Voir Fixation de l’appareil)
7. Procéder aux mesurages.
(Voir Mesurer)
8. Voir les mesures.
(Voir Afficher les mesures etTransférer les données sur PC)

NOTA : Consulter les consignes de sécurité dans le Manuel de l’Utilisateur du 4448.
37

Francais

A la réception

Manipulation des boutons











Manipulation des boutons

 Bouton gauche Bouton droit Indicateur d’état Display Windshield Capteur IR Sensor

Pour recharger le 4448

Pour relier plusieurs chargeurs

Placer l’appareil sur le chargeur. La
recharge complète d’une batterieépuisée prend environ 90 minutes.

Relier les différents chargeurs au
moyen du Câble AO-1493 puis
connecter l’Alimentation ZG-0864
à une entrée AC disponible sur unquelconque de ces chargeurs.
Quatre chargeurs peuvent ainsi êtrealimentés à la fois.

Chargement en cours : LED rougeclignotante
Chargement terminé : LED bleue
constante
S’assurer que le 4448 asuffisamment d’autonomie
pour la durée demesurage souhaitée en
vérifiant sur le display ladurée disponible
affichée.

Francais

Vue synoptique Presser une fois

 Maintenir enfoncé

Presser plusieurs fois
(ici, trois fois)

38

39

Pour mettre le 4448 en marchePresser le bouton de gauche (L). Le 4448 lance
une séquence de démarrage.
Le premier affichage montre le numéro dumodèle et la version du firmware. Le second
affichage montre un texte défini par l’utilisateur(qui peut être configuré via le Logiciel 7825
Protector).
L’appareil passe ensuite en mode de veille,
affichant alternativement la date et l’heure et(quand elles existent) les données précédemment
recueillies.

Pour mettre le 4448 à l’arrêt
Maintenir le bouton ‘L’ enfoncé pendant 3
secondes.


3 sec

40

Configurer au démarrage
Pour passer en mode
Configuration

Pour changer la valeur
Presser le bouton ‘R’pendant la séquencede démarrage et lemaintenir enfoncé.

Paramètres
CONTRASTLANGUAGE
PRO. MODEDISPLAY MODE
dB ALARMS
PRESET MODE
AUTO
CLEAR MEMORY

Presser le bouton ‘L’pour changer la valeur
du paramètre.

Pour choisir leParamètre



Presser le bouton ‘R’pour confirmer la valeur
et passer au paramètresuivant.
Après confirmation du dernierparamètre, l’appareil
revient en mode defonctionnement normal.

NOTA : Si la mémoire doit être vidée, un second
affichage apparaît demandant confirmation. Presser
le bouton ‘L’ pour afficher ‘Yes’, puis le bouton ‘R’pour confirmer l’effacement.
41

Francais

Commutateur de marche/arrêt

Calibrage

Boule antivent
Pour déposer la boule
antivent

NOTA : Par défaut, le 4448
est réglé pour un niveau decalibrage de 94dB. Ce réglage
peut être modifié au moyen du
Logiciel sur PC 7825 Protector(pour information, consulter
l’aide en ligne du 7825 etle Manuel de l’Utilisateur du4448).

Tourner vers la gauche labague en plastique noir à labase de la boule (pas la partie
en mousse) et la soulever.

Pour fixer la boule antivent

1. Mettre l’appareil en marche etattendre qu’il ait démarré.
2. Retirer la boule antivent(Important : voir Boule antivent)
3. Placer le calibreur acoustique surle microphone, comme montré àgauche. Vérifier que le calibreur
est bien positionné sur le 4448,
sans torsion qui pourrait dévisser
ou trop visser la capsule dumicrophone.

42

4. Mettre le calibreur en marche. Le
4448 reconnaît automatiquement
la tonalité de calibrage et passe enmode de calibrage.
5. Appuyer sur le bouton ‘R’ pourconfirmer le calibrage. Le 4448
s’ajuste automatiquement surle niveau requis après quelques
secondes.
Le message ‘CAL OK’ marque le terme
du calibrage. Le 4448 stock heure, dateet niveau, puis revient à l’écran pardéfaut.

Placer les ergots dans les fentes
puis tourner doucement vers
la droite la bague en plastiquenoir à la base de la boule (pasla partie en mousse).

Francais

Pour calibrer l’appareil

NOTA : Pour ces opérations, bien s’assurer de manœuvrer la bague en plastique noir àla base de la boule, jamais la partie en mousse

43

Pour fixer l’appareil sur la personne
1. Choisir le dispositif de fixation
adéquat à l’application (activité
physique, environnement, vêtement
de travail exigé). Par exemple, unebroche assure une bonne fixation,
mais troue les tissus protecteurs.
2. Placer l’appareil sur l’épaule ens’assurant que le microphone pointe
vers le haut et qu’il est 10 cm environ
de l’oreille.
Pour un supplément d’information, sereporter au Manuel de l’Utilisateur.

Changer les pinces de fixation
1. Localiser et déposer les vis (voirillustration) et enlever les pinces.
2. Placer les ergots des nouvelles
pinces dans les trous, positionner
délicatement les pinces, puis remettre
les vis et serrer celles-ci.

Mesurer
Pour lancer un mesurage


3 sec





3 sec

Maintenir les deux boutons enfoncés
pendant 3 secondes ; un compte àrebours s’affiche sur le display.
En cours de mesurage, le symbole► apparaît en haut et à gauche
du display. Sont alors affichés enboucle : le niveau en dB, la durée dumesurage, et l’état de la batterie/mémoire.

Pour stopper un mesurage
Maintenir les deux boutons enfoncés
pendant 3 secondes ; un compte àrebours s’affiche sur le display.

44

Verrouiller/déverrouiller boutons
et display pendant le mesurage

Maintenir le bouton ‘R’ enfoncéet presser 3 fois le bouton ‘L’. Un
cadenas s’affiche sur le display. Les
boutons sont maintenant verrouillés
et les valeurs dB n’apparaissent pas.
Pour déverrouiller, maintenir ‘R’enfoncé et presser ‘L’ 3 fois.

Francais

Fixation de l’appareil

Durée de mesurage
préprogrammée
L’appareil peut aussi être verrouillé(optionnel) lors d’un mesurage
préprogrammé. Le mesurages’arrêtera à l’heure prédéfinie. Pour
stopper le mesurage avant terme,déverrouiller le 4448 comme décrit
ci-dessus.
45

Transférer les données sur PC

Quand le 4448 ne mesure pas, le display
affiche en boucle les données du mesurage
précédent. Celles-ci dépendent de la configuration du 4448 (données ISO ou OSHA). Voir àgauche les données affichées.

ISO

OSHA

10Hrs
MEMOIRE 80Hrs

10Hrs
MEMOIRE 80Hrs

 13:45:03

 13:45:03

DUREE
07:45:12

DUREE
07:45:12

LAEQ 89.9
LCPK 101.4  dB

LAVG 111.4
LZPK 119.4  dB

PA2Hrs

3.20

OSHA DOSE
4.3  %PROJ DOSE
352.5  %PROJ DOSE
352.5  %

LAF 112.3  dB

46

Pour geler l’affichage

LAS 109.3  dB

Presser le bouton ‘R’ pour interrompre à tout
moment l’affichage en boucle pendant 10
secondes.

Transfert des données
1. Vérifier que l’adaptateur
infrarouge (AO-1492) est relié auport USB du PC et que la batterie
du 4448 est suffisamment chargée.
2. Démarrer le Logiciel 7825 Protector
sur le PC, ainsi que le 4448.
3. Placer l’interface infrarouge du4448 face à l’adaptateur infrarougedu PC.
4. Lancer la procédure de transfert àpartir du 7825 Protector et suivre
les instructions qui apparaissent.
Francais

Afficher les mesures

Messages d’erreur
Si un message d’erreur apparaît aucours d’une des procédures décrites
dans ce Guide pratique, consulter lechapitre Dépannage du Manuel del’Utilisateur du 4448.
Une LED rouge clignotante indiqueque la batterie ou la capacité demémoire est faible.

47

Spécifications
AUTONOMIE
28 heures

CAPACITÉ MÉMOIRE/ HISTORIQUE
180 heures (enregistrements de 1 minute)

DURÉE DE MESURAGE PRÉPROGRAMMABLE
00:00–12:00 par pas de 30 minutes

GAMME DE MESURAGE
Gamme de fonctionnement linéaire :Gamme de mesurage crête :65,0 – 140,3dB (Eff)95,0 – 143,3dB (Crête)TEMPÉRATURES DE FONCTIONNEMENT
0 à +40°C

HUMIDITÉ
30% – 95% RH (sans condensation)48

1. Controllo contenuto e tasti simbolo.
(Generalità)
2. Piena carica delle unità 4448.
(Carica)
3. Accensione delle unità.
(Accensione e spegnimento)
4. Configurazione delle unità sui parametri
d’uso desiderati.
(Configurzione all’avvio)
5. Calibrazione delle unità.
(Calibrazione)
6. Applicazione delle unità sui soggetti.
Opzionale: sostituzione delle clip difissaggio. (Montaggio)
7. Esecuzione della misura.
(Controllo misura)
8. Accesso e revisione dei dati.
(Revisione dati di misura e Scarico
dati a PC)

NOTA: per informazioni sulla sicurezza, consultare il Manuale utente del tipo 4448.
49

Italiano

Primo uso fuori dalla confezione

Generalità

Carica









Tasti simbolo

Per collegare i caricatori

Collocare l’unità nell’apposito
caricabatteria. Il caricamento completorichiederà circa 90 minuti.
In carica: luce rossa intermittente.
Carica terminata: luce blu costante.
Accertarsi che il 4448 abbia abbastanza
carica per la durata richiesta dellamisura controllando lo schermo che
indicherà il tempo residuo di funzionamento disponibile.

Collegare i caricabatteria usando ilcavo
AO-1493 e collegare l’alimentatore
ZG-0864 a un caricatore con uningresso AC libero (qualsiasi ingresso
AC andrà bene).
L’alimentatore può alimentare una
catena che comprende fino a quattro
caricatori.

 Tasto sinistro Tasto destro
 Indicatore stato Display Schermo antivento Sensore infrarosso

ItalianoPer caricare il tipo 4448

 Premere una volta

 Premere e tenere premuto
Premere più volte(nell’esempio, tre volte)

5051Per accendere il tipo 4448

Premere il tasto sinistro (L). Il 4448 eseguirà una
sequenza di avviamento.
Il primo schermo visualizza il numero di modelloe la versione firmware. Il secondo schermo
mostra il testo definito dall’utente (che può
essere configurato usando il programma
Protector 7825).
Dopo l’avviamento lo strumento entra inmodalità standby, alternando la data e l’ora e (sedisponibile) la data dell’ultima esecuzione.

Per spegnere il tipo 4448
Mantenere premuto il tasto ‘L’ per 3 secondi.


3 sec

52

Configurazione all’avvio
Per immettere la modalità Configurazione

Per modificare ilvalorePremere e mantenere
premuto il tasto
destro ‘R’ durante
l’avviamento.

Parametri
CONTRASTO
LINGUA
MODO PRO.
MODO DISPLAY
ALLARMI dB
MODO PRESELAUTOCANCELLA MEMORIAPremere il tasto ‘L’ permodificare il valore delparametro.

Per selezionare ilparametroPremere il tasto ‘R’ perconfermare il valore e
passare al parametro
successivo.
Confermando lo
schermo finale, lo
strumento torna
al funzionamento
standard.

Italiano

Accensione e spegnimento

NOTA: se si richiede la cancellazione della memoria,
apparirà un secondo schermo con la richiesta diconferma. Cambiare l’opzione su ‘Sì’, premendo iltasto ‘L’, e premere ‘R’ per confermare.
53

Calibrazione

Schermo antivento
Per smontare lo schermo
antivento

NOTA: inizialmente il tipo
4448 è impostato per un livellodi calibrazione di 94dB. È
possible modificarlo usando il
software per PC Protector 7825
(per ulteriori dettagli, si consultil’Help on-line del 7825 e ilmanuale utente del 4448).

Girare l’anello di plasticanero (non la spugna) in senso
antiorario e sollevarlo.

Per montare lo schermo
antivento

1. Accendere il 4448 e attendere il
suo avviamento.
2. Smontare lo schermo antivento(Importante: Schermo
antivento)
3. Posizionare il calibratore acustico
sul microfono come mostrato asinistra. Fare attenzione a premere
il calibratore senza girare, perchè
potrebbe allentare o stringere
troppo la capsula del microfono.

54

4. Accendere il calibratore. Il 4448
riconosce automaticamente i
toni e entra nella modalità dicalibrazione.
5. Premere il tasto ‘R’ per confermare
l’esecuzione della calibrazione.
Occorreranno pochi secondi
allo strumento per correggersi
automaticamente al livellorichiesto.
Una volta completata, apparirà
il messaggio ‘CAL OK’. Il 4448
memorizza la data, l’ora ed il livellodi calibrazione, ritornando poi allo
schermo iniziale.

Posizionare le linguette nelle
asole e girare cautamentel’anello di plastica nero (non la
spugna) in senso orario.

Italiano

Per calibrare il tipo 4448

NOTA: smontando o montando lo schermo antivento si raccomanda di usare e girare
solo l’anello di plastica nero e non la spugna.

55Per montare le unità
1. Selezionare le clip adatteall’applicazione (movimento fisicopiù o meno intenso, ambiente erichieste individuali di abbigliamentodel soggetto). Per esempio, gli spilli
forniscono un aggancio più solido mapenetrano nei materiali protettivi.
2. Posizionare lo strumento sulla
spalla accertandosi che il microfono
sia orientato verso l’alto e che sia
posizionato a circa 10 cm dall’orecchio.
Per ulteriori informazioni sul montaggio,
si veda il manuale.

Per rimuovere o sostituire le clip dimontaggio
1. Individuare e rimuovere le viti (vedere
l’immagine) e sollevare le clip.
2. Posizionare le linguette delle nuove
clip nei fori, premere leggermente inposizione, poi inserire e stringere le
viti.

56

Controllo misura
Per avviare la misura


3 sec

Per bloccare e sbloccare i tasti e il
display durante la misura


3 sec

Premere e mantenere premuti
entrambi i tasti per 3 secondi; il
conto alla rovescia sarà indicato nello
schermo.
Durante una misura, sullo schermo inalto a sinistra appare il segnale ►. Lo
schermo visualizzerà poi alternandoli,
il livello in dB, la durata della misurae lo stato della memoria/batteria.

Per arrestare la misura
Premere e mantere premuti entrambi
i tasti per 3 secondi; il conto alla
rovescia sarà indicato nello schermo.


Tenere premuto ‘R’ e premere ‘L’
per 3 volte. Un simbolo a lucchetto
appare nello schermo ad indicare
che i tasti sono bloccati e i valori indB verranno nascosti. Per sbloccare i
tasti, mantenere premuto il tasto ‘R’ epremere il tasto ‘L’ per 3 volte.
ItalianoMontaggio

Preselezione del tempo di misura
Con un tempo di misurapreselezionato, durante la misura,l’unità può entrare in modalitàbloccata (opzionale). La misura si
arresta al tempo impostato.
Per arrestare la misura prima,
sbloccare lo strumento come sopra
descritto.
57

Scarico dati a PC

Quando il 4448 non sta misurando, lo
schermo visualizza, alternando, informazioni
e dati della misura precedentemente
eseguita. I dati visualizzati dipendeno dalla
configurazione del 4448: dati ISO o OSHA. I
dati mostrati sono visualizzati qui a sinistra.

ISO

OSHA

10Hrs
MEMORIA 80Hrs

10Hrs
MEMORIA 80Hrs

 13:45:03

 13:45:03

DURATA
07:45:12

DURATA
07:45:12

LAEQ 89.9
LCPK 101.4  dB

LAVG 111.4
LZPK 119.4  dB

PA2Hrs

3.20

OSHA DOSE
4.3  %PROJ DOSE
352.5  %PROJ DOSE
352.5  %

LAF 112.3  dB

58

LAS 109.3  dB

Per congelare il DisplayPremere il tasto ‘R’ su qualsiasi di queste
schermate per congelare il display per circa
10 secondi, per poi riprendere.

Per scaricare i dati ad un PC
1. Accertarsi che l’adattatore di scarico
a infrarossi (AO-1492) sia collegato
alla porta USB del PC e che il 4448
sia sufficientemente in carica.
2. Avviare il software Protector 7825
nel PC e il 4448.
3. Mantenere la porta a infrarosssidel 4448 vicino e orientata verso
l’adattatore a infrarossi del PC.
4. Iniziare la sequenza di scarico
con il Protector 7825 e seguire leistruzioni per il caricamento dei
dati.

Messaggi d’errore
Se durante le procedure descritte
dovesse apparire un messaggio
d’errore, si raccomanda di riferirsi allasezione Individuazione guasti, nel
Manuale utente del tipo 4448.
Un LED rosso intermittente indica chela batteria è scarica, o lo spazio dellamemoria è esaurito.

Italiano

Revisione dati di misura

59

Dati tecnici
DURATA BATTERIA
28 oreCRONOLOGIA / CAPACITÀ MEMORIA
180 ore con periodo registrazione di 1 minuto

TEMPO DI MISURA PRESELEZIONATO
00:00–12:00 in intervalli di 30 minuti

GAMMA DI MISURA
Gamma di funzionamento lineare:
Gamma di misura del picco:

65.0 – 140.3dB (RMS)
95.0 – 143.3dB (Picco)

TEMPERATURA DI FUNZIONAMENTO
0 a +40°C

UMIDITÀ
30% – 95% RH (non condensante)60Primeiro Uso fora da Caixa

Portuguese

1. Verifique o conteúdo e Símbolo da
tecla. (Veja Resumo)
2. Carregar totalmente Tipo 4448 unidade(s). (Veja Carga)
3. Ligue o aparelho (s)(Veja Ligar e Desligar)
4. Configurar unidade (s) aos parâmetrosde uso desejado.
(Veja Configurar a inicialização)5. Calibrar unidade (s).
(Veja Calibrar)
6. Montagem das unidades. Opcional:
Alterar os clips (Veja Montegem)
7. Realizar as medições.
(Veja Controle de Medição)8. Acesso e revisão de seus dados.
(Veja Revisão dados de medição e
Download de dados para o PC)

NOTA: Consulte Tipo 4448 - Manual do Usuário para informações de segurança.
61Visão Geral

Carga









Símbolo da tecla

Para Ligar os Carregadores

Coloque a unidade sobre o carregador
fornecido. O carregamento total doplano levará aproximadamente 90
minutos.
Carregando: a luz vermelha piscando.
Carregado: luz azul constante.
Certifique que o Tipo 4448 tem cargasuficiente para duração da medição
olhando no display, que irá
mostrar o tempo disponívelpara a corrida para
carga atual.

Você pode ligar vários carregadoresem paralelo, ligue os carregadores
usando o cabo AO-1493 em seguida
ligar a fonte de alimentação ZG-0864 a
um carregador com entrada livre AC.
A fonte de alimentação pode conduzir
uma corrente para até quatro
carregadores.

 Tecla esquerda
 Tecla direita
 Indicador de Status Display Windshield (protetor de vento e poeira) Sensor Infravermelho Pressione uma vez

PortuguesePara Carga Tipo 4448

 Pressione e segure

Pressione várias vezes(exemplo: mostra três vezes)

6263Configuração na inicialização
Para ligar Tipo 4448

Pressione a tecla esquerda (L). Tipo de 4448
será executado através de uma seqüência deinicialização.
A primeira tela exibe o número do modelo eversão do firmware. A segunda tela mostra
o texto definido pelo usuário (pode serconfigurado através Software Tipo Protector
7825).
Depois de iniciar o instrumento entra em modo
stand-by, alternadamente montrando a horae a data e (se disponível) os dados da última
execução.

Para desligar o Tipo 4448
Mantenha pressionada a tecla esquerda (L) por
3 segundos.


3 sec

64

Para ativar o modo deconfiguração

Para alterar o valor
Pressione e segure atecla ‘R’ durante oarranque.

Parâmetros
CONTRASTE
LÍNGUA
PRO. MODODISPLAY MODO
dB ALARMESPRESET MODOAUTOAPAGAR MEMÓRIA

Aperte a tecla ‘L’ paraalterar o valor doparâmetro.

Para selecionar oparâmetro

Aperte a tecla ‘R’ para
confirmar o valor e
passar para o próximo
parâmetro.
Confirmando a tela final
retorno da unidade à
operação padrão.

NOTA: Se a memória deve ser apagada, uma segunda
tela será exibida para confirmação. Altere para opção
‘Yes’ (Sim) pressionando a tecla ‘L’ , em seguidapressione a tecla ‘R’ para confirmação e eliminação.

Portuguese

Ligar e Desligar

65

Calibração

Windshield
Para remover o WindshieldRode o anel de plástico preto(não a espuma) anti-horário eretire-a.

NOTA: Por padrão Tipo 4448
está configurado para um nível
de calibração de 94dB. Vocêpode mudar isso usando o
Software para PC Protector
7825 (para mais detalhes veja
Tipo 7825 Ajuda on-line e Tipo
4448 Manual do Usuário).

Para fixar (prender) oWindshield

1. Ligue Tipo 4448 e esperar para que
inície2. Remova windshield(Importante: Veja Windshield)
3. Coloque o calibrador acústico
sobre o microfone como mostrado
à esquerda.Tenha cuidado parapressionar o calibrador em linha
reta em Type 4448 sem torcer, pois
isso pode afrouxar ou apertar omicrofone.Turn on the calibrator.
Type 4448 automatically recognises
calibration tones and enterscalibration mode (see left).
66

4. Aperte a tecla ‘R’ para confirmar
que deseja calibrar. Tipo 4448 vai
demorar alguns segundos para
ajustar automaticamente para onível exigido.
Uma vez que a calibração é concluida,
uma mensagem “CAL OK” aparecerá.
Tipo 4448 armazena a data decalibração, a hora e o nível, em
seguida retorna às telas padrão.

Coloque as presilhas nos
encaixes, em seguida, giresuavemente o anel de plásticopreto (não a espuma) no
sentido horário.

NOTA: Para colocar ou remover o windshield sempre mantenha a torção usando o anelde plástico preto na base do windshield: NUNCA na espuma.

67

Portuguese

Para Calibrar o Tipo 4448

Para montar as unidades
1. Selecione os clips corretos para aaplicação (mais ou menos intensa
atividade física, meio ambiente e
sujeito às exigências de vestuário). Por
exemplo, alfinetes para prender mais
forte e penetrar materiais de proteção
– Jaleco de couro
2. Coloque o instrumento sobre o ombro
para garantir que o microfone fiquepara cima e que a ponta do microfone
é de cerca de 10 centímetros daorelha.
Para mais informações de montagem
consulte o manual.

Para Remover ou Mudar clips demontagem
1. Localize e remova os parafusos (verimagem) e levante os clips.
2. Coloque as abas dos novos clips nos
buracos, empurre os clips no lugar e
insira e aperte os parafusos.

68

Controle de Medição
Para iniciar uma medição


3 sec


3 sec

Pressione e segure as duas teclas por
3 segundos; uma contagem regressiva
sera apresentada na tela.
Durante a medição, ► será exibido
no canto superior esquerdo da tela.
A tela irá alternar entre nível dB,
duração da medição, e a memória/status da bateria.
Para para uma medição
Pressione e segure as duas teclas por
3 segundos; uma contagem regressiva
será apresentada na tela.

Para bloquear e desbloquear teclas
e visor durante a medição


Segure a tecla “R“ e pressionar a tecla‘L’ 3 vezes. O símbolo de cadeadoaparecerá. As teclas são bloqueadas, eos valores dB serão escondidos.
Para desbloquear as teclas, segurea tecla “R” e pressionar a tecla ‘L’ 3
vezes.
Pré-ajuste do tempo de medição
Ao usar o tempo de medição
pré-ajuste, a unidade pode entrar no
modo bloqueado (opcional) durante
a medição. A medida vai parar no
prazo estabelecido.
Para parar a medição com
antecedência, desbloquear Tipo 4448
conforme descrito acima.
69

Portuguese

MontarRever dados de Medição

Download Dados para PC

Quando o Tipo 4448 não está medindo, atela irá percorrer as informações de status eos dados da corrida de medição anterior. Os
dados apresentados dependerá da configuração do tipo 4448, os dados NR15, ISO ou
OSHA. Os dados apresentados é exibido àesquerda.

OSHA

10Hrs
MEMORIA 80Hrs

10Hrs
MEMORIA 80Hrs

 13:45:03

 13:45:03

DURACAO
07:45:12

DURACAO
07:45:12

LAEQ 89.9
LCPK 101.4  dB

LAVG 111.4
LZPK 119.4  dB

PA2Hrs

3.20

OSHA DOSE
4.3  %PROJ DOSE
352.5  %PROJ DOSE
352.5  %

LAF 112.3  dB

70

LAS 109.3  dB

Para congelar a exibição



1. Verifique se o adaptador de
download de infra-vermelhos(AO-1492) é conectadoà porta USB
do seu PC e 4448 Tipo tem cargasuficiente.
2. Inicie Tipo 7825 Protector softwareno PC e Tipo 4448.
3. Segure Tipo 4448’s infra-vermelho
proximo a porta e de frente para oadaptador infravermelho no PC.
4. Iniciar sequencia de download
dentro Tipo 7825 Protector e siga as
instruções para fazer o download
de dados.

Mensagens de erro

Aperte a tecla ‘R’ em qualquer uma dessas
telas e a tela irá congelar por 10 segundos edepois continuar.

Se alguma mensagem de erro ocorrer
durante os procedimentos descritos
neste guia de campo, consulte“Solucionando problemas no Tipo4448” no Manual do Usuário.
Um LED vermelho piscando indica
bateria fraca ou memória.

Portuguese

ISO

Para transferir dados para um PC

71

Especificações de CampoDURAÇÃO DA BATERIA
28 horas

TEMPO DE MEDIÇÃO / CAPACIDADE DE MEMÓRIA
180 horas com 1 minuto de integração

PRESET TEMPO DE MEDIÇÃO
00:00–12:00 em etapas de 30 minutos

RANGE DE MEDIÇÃO
Range de Operação Linear:
Range de Medição Peak

65.0 – 140.3dB (RMS)
95.0 – 143.3dB (Peak)

TEMPERATURA DE OPERAÇÃO
0 a +40°C

HUMIDADE

72

BR  2212-11

30% – 95% RH (sem condensação)HEADQUARTERS: Brüel & Kjær Sound & Vibration Measurement A/S · DK-2850 Nærum · Denmark
Telephone: +45 77 41 20 00 · Fax: +45 45 80 14 05 · www.bksv.com · info@bksv.com
Local representatives and service organisations worldwide