UCL / SFC011C
クランプオン型超音波流量計

IM-F2046-J00

取扱説明書

初版 2020 10 E

目次

■ 本書で使用しているマークについて........................................................................................................... 3■ 使用上の一般的注意事項 ............................................................................................................................ 3■ 標準仕様 ..................................................................................................................................................... 41. 設置方法 ..................................................................................................................................................... 61.1 検出器の設置............................................................................................................................................. 61.1.1 設置場所............................................................................................................................................. 61.1.2 配管への取付位置 .............................................................................................................................. 61.1.3 配管への取付手順（小口径） ............................................................................................................ 71.1.4 配管への取付手順（大口径） ............................................................................................................ 81.2 変換器の設置............................................................................................................................................. 91.2.1 設置場所............................................................................................................................................. 91.2.2 取付および取外し方法 ....................................................................................................................... 91.2.3 設置方向について .............................................................................................................................. 91.3 配線 ......................................................................................................................................................... 101.3.1 検出器の配線 ................................................................................................................................... 101.3.2 変換器の配線 ................................................................................................................................... 102. パネル部の名称と機能 .............................................................................................................................. 122.1 名称と機能 .............................................................................................................................................. 122.1.1 ゼロ点調整スイッチ....................................................................................................................... 122.1.2 アドレススイッチ .......................................................................................................................... 132.1.3 表示器 ............................................................................................................................................ 132.1.3.1 表示内容 .................................................................................................................................................... 132.1.3.2 表示動作 .................................................................................................................................................... 153. 運転 .......................................................................................................................................................... 163.1 運転準備 .................................................................................................................................................. 163.1.1 電源投入前の確認事項 ..................................................................................................................... 163.1.2 検出器通水 ....................................................................................................................................... 163.2 運転 ......................................................................................................................................................... 163.2.1 通電 .................................................................................................................................................. 163.2.2 運転 .................................................................................................................................................. 163.2.3 ゼロ点の調整 ................................................................................................................................... 164. パラメータ ................................................................................................................................................ 175. 通信関連 ................................................................................................................................................... 175.1 パラメータ設定 ....................................................................................................................................... 175.2 通信仕様 ................................................................................................................................................ 175.3 アドレス設定 ......................................................................................................................................... 176. 日常点検 ................................................................................................................................................... 186.1 接続配管の点検 ....................................................................................................................................... 186.2 防水性の点検........................................................................................................................................... 186.3 検出器の点検........................................................................................................................................... 18
7. トラブル一覧 ............................................................................................................................................ 182IM-F2046-J00

■ 本書で使用しているマークについて
本書では、安全上絶対にしないでいただきたいことや注意していただきたいこと、また、取扱い上守っていただきたいことの説明に次のようなマークを付けています。これらのマークの箇所は必ずお読み下さい。

警告

この表示を無視して誤った取扱いをすると、使用者が死亡または重傷を負う可能性が想定される内容を示しています。

注意

この表示を無視して誤った取り扱いをすると、使用者が傷害を負う可能性や製品の破損または付帯設備等の物的損害の発生が想定される内容を示します。

注記

この表示は製品の取り扱い上、必要不可欠な操作や情報を示しています。

■ 使用上の一般的注意事項

警告

警告

警告

警告

改造等の禁止本製品は工業用計器として厳密な品質管理のもとに製造・調整・検査を行い納入しております。
みだりに改造や変更を行うと本来の性能を発揮できないばかりか、作動不適合や事故の原因となります。改造や変更は行わないで下さい。
仕様変更の必要がある場合は当社までご連絡下さい。

注意

使用条件の厳守
納入仕様書あるいはテクニカルガイ
ダンスに記載された仕様、圧力、温度の範囲内での使用を厳守して下さい。
この範囲を超えた条件での使用は事故、故障、破損などの原因となります。

警告

用途
本製品は計器としての用途にのみ使用し、その他の用途には使用しないで下さい。

材 質
本製品の材質については納入仕様書あるいはテクニカルガイダンスに記載されています。当社でもお客様の仕様をお伺いし最適な材質選定に努めておりますが、実際のプロセスにおいては混入物などの影響があり、耐食性等が万全でないこともあります。
耐食性・適合性のご確認、最終的な材質の決定はお客様の責任でお願いいたします。

制御上の安全性
本製品は工業計器として最善の品質管理のもとに製造、調査、検査を行い納入いたしておりますが、各種の原因で不測の故障が発生する可能性もあります。安全上の重大な問題が発生する可能性のあるプロセスコントロールなど
において本製品を使用する場合は、万一に備えて本製品に加えて同様な機能を果たす機器を併設し、二重化を行う
ことにより一層の安全性を確保して下さい。

保守・点検
本製品を保守、点検などのためにプロセスから取り外す際は、測定対象物の計器内への付着に注意して下さい。
測定対象物に腐食性や毒性がある場
合は、作業者に危険がおよびます。

3IM-F2046-J00

■標準仕様■ 検出器 UCL 仕様■ 変換器 SFC011C 仕様

測定対象
流体温度
周囲温度
流体圧力
流体音速
流体動粘度
構造
流量範囲

電源、I/O 仕様
電源電圧
消費電流
突入電流
表示器
デジタル出力

分類

小口径

大口径

：液体全般（気泡を含まない液体）：10～60℃：0～60℃：0～0.5MPa：1000～2200 m/s：0.8～40.0 mm2/s：IP64 相当（屋内仕様、配管取付時）：表１参照
表 1. 流量範囲
チューブサイズ(外径[mm]×内径[mm])
φ6mm (6×4)
φ8mm (8×6)
φ10mm (10×8)
φ1/4”(6.35×4.35)
φ3/8”(9.53×6.38)
φ1/2”(12.70×9.55)
φ3/4”(19.05×15.90)
φ1”(25.40×22.25)

アナログ出力流量範囲[L/min]0 ～ 3.00 ～ 8.00 ～ 8.00 ～ 3.00 ～ 8.00 ～ 20.0
0 ～ 50.0
0 ～ 80.0

通信仕様：DC24V±10%：約 350mA：約 800mA：4 桁（瞬時流量、ステータス）：周波数、積算、アラーム、エラーから選択
オープンコレクタ、デューティ 1:1
負荷抵抗 DC30V、10mA 以内：DC4-20mA（標準）負荷抵抗 500Ω以下※形式により出力の種類を選択可能です。
：RS-485 半二重通信調歩同期：Modbus Protocol、RTU mode
：Baud Rate
57.6kbps：Data size
8bits
：Parity
Even：Stop bit
1bit
：Address Switch 1～32機能、構造仕様
本体質量：小口径 約 120g（ケーブル長 0.2m）
大口径 約 90g（ケーブル長 0.2m）
中継ケーブル質量：約 350g
本体材質
：PPS（大口径クランプバンド：PP）
本体ケーブル材質：PTFE 被覆
中継ケーブル材質：PVC 被覆

パラメータ設定
周囲温度
周囲湿度
取付
構
造
ケース材質、色質
量
適合検出器
適合規格

4：専用コンフィグソフトによる設定
又はメモリ付検出器より読込み：単独:0～45℃、連結:0～25℃：30～80%RH（結露しないこと）：DIN レール取付：IP20 相当（屋内仕様）：耐熱 ABS 樹脂（白）：約 250g（給電端子込み）：UCL シリーズ：EMC 適合規格：EN61326-1、EN61326-2-3
RoHS2 対応

IM-F2046-J00

外形寸法図■ 変換器 SFC011C■ 検出器 UCL
小口径

大口径

表 2. 検出器寸法
分類小口径

大口径

チューブサイズ
φ6mm
φ8mm
φ10mm
φ1/4”
φ3/8”
φ1/2”
φ3/4”
φ1”
5寸法 [mm]DA
68106.35
9.53
12.70(65)
19.05
(71)
25.40(78)IM-F2046-J00

1. 設置方法
1.1 検出器の設置
1.1.1 設置場所

注記

設置場所は下記の条件を考慮して下さい。
1) 周囲温度が 0℃～60℃で、直射日光の当たらない屋内。
急激な温度の変化がないところをお勧めします。
2) 誘導障害を受ける恐れのない場所。動力機器の近くなどは避けて下さい。
3) 水滴や、腐食性ガスのない所。
4) 保守点検が容易にできる場所。

1.1.2 配管への取付位置

注記

正しい測定を行うために、次の項目について考慮して取付位置の選定および取付を行って下さい。
1) 測定管内が常に流体で満たされている事。
水平、垂直、斜めの配管いずれでも取り付けできますが、気泡が溜まりにくい垂直配管への設置をお奨めします。
水平配管に設置する場合は、小口径・大口径共に気泡の影響を受けにくい下記取付姿勢をお奨めします。
【小口径】

推奨取付（垂直配管）

超音波信号気体推奨取付液体（水平配管）【大口径】

推奨取付（垂直配管）
超音波信号
推奨取付
気体

（水平配管）

液体

2)
3)
4)

安定して測定するために、上流側に内径の 10 倍、下流側に内径の 5 倍以上の直管長を設けるようにして下さい。
検出器には流れ方向を矢印で示していますので、流れ方向は必ず矢印の向きに合うように取り付けて下さい。
（1.1.3 の 3）、1.1.4 の 1）参照）
配管応力がないように取付けて下さい。
6

IM-F2046-J00

1.1.3 配管への取付手順（小口径）

注記

1)

検出器のストッパーを 90°回転して外します。

2)

検出器のクランプ部を広げます。

3)

接続するチューブを挟み、クランプ部を閉じます。この際、流れ方向と矢印の方向を合わせて下さい。

矢印

4)

検出器のストッパーを取付けて完了です。

※取付状態を安定させるため、取付後 60 分以上経過してから使用（ゼロ点調節）することを推奨します。
7IM-F2046-J00

1.1.4 配管への取付手順（大口径）1)

注記接続するチューブに検出器を挟み込みます。この際、流れ方向と矢印の方向を合わせて下さい。

矢印2)

クランプバンドの一方の端部を他方の端部に引っ掛け、つまみを寝かせてロックします。

⇒⇒

3) クランプバンドを 2 ヵ所取付ければ完了です。

※取付状態を安定させるため、取付後 60 分以上経過してから使用（ゼロ点調節）することを推奨します。

8IM-F2046-J00

1.2 変換器の設置
注記

1.2.1 設置場所

設置場所は下記の条件を考慮して下さい。
1) 周囲温度が “単独時：0℃～45℃”、“連結時：0℃～25℃”で直射日光の当たらない屋内。
2) 屋内周囲湿度 30～80%RH（結露しないこと）。
3) 誘導障害を受ける恐れのない場所。動力機器の近くなどは避けて下さい。
4) 水没する恐れのない場所。

1.2.2 設置および取外し方法
1)2)

注記変換器は DIN レール設置タイプで、簡単に配線工事を行えます。
変換器本体が DIN レールにしっかり固定されていることを確認して下さい。

注記

1.2.3 設置方向について設置方向は②、③、④を推奨します。①の場合は 1 台のみでご使用下さい。
設置方向①設置方向②上側

上側

上側

設置方向④設置方向③上側

9IM-F2046-J00

1.3 配線

注記

表 3. 変換器配線
No.
①

名称
給電端子

②

保護カバー

③センサ
コネクタ

内容
電源や通信線を接続します。
変換器を連結して使用する場合、末端の変換器に取付けます。
検出器を接続します。
6…CH6
5…CH5
4…CH4
3…CH3
2…CH2
1…CH1

1.3.1 検出器の配線検出器と変換器の接続には中継ケーブルを使用します。中継ケーブルは検出器の中継端子、変換器のセンサコネクタにそれぞれ接続して配線します。その際、使用する CH を間違えないようにして下さい。

中継ケーブルコネクタ中継端子

1.3.2 変換器の配線
給電端子

表 4. 給電端子配線
No.
123456極性+FG+
SG

名称
電源（DC24V）

通信（RS-485）

10IM-F2046-J00

I/O 端子表 5. デジタル出力端子配線
デジタル出力端子No.
12345678910111213～24アナログ出力端子

表 6. アナログ出力端子配線

内容
CH1 出力+CH1 出力CH2 出力+
CH2 出力CH3 出力+
CH3 出力CH4 出力+
CH4 出力CH5 出力+
CH5 出力CH6 出力+
CH6 出力未使用

No.
123456789101112

内容
CH1 出力+CH1 出力CH2 出力+
CH2 出力CH3 出力+
CH3 出力CH4 出力+
CH4 出力CH5 出力+
CH5 出力CH6 出力+
CH6 出力-※デジタル出力は周波数（0-1kHz）、アラーム、エラー出力の中から選択します。
※アナログ出力は 0-20mA、4-20mA、0-5V、1-5V の中から選択します。

1)

電源の接続
・ 変換器の電源入力は専用コネクタで接続して下さい。
・ 銘板上に記載されている電源電圧と、接続しようとする電源電圧が一致していることを確認して下さい。
・ 電源は必ず計装用電源などを使用し、動力用電源と共有することは避けて下さい。
・ 電源端子にて、電源電圧が規定範囲内にあることを確認して下さい。
・ 極性にご注意下さい。
・ 変換器の消費電流は約 350mA です。使用する台数分の電流を確保して下さい。
・ SFC011C は連結コネクタを内蔵しており、これを用いることで複数台の連結が可能です。
・ 連結型コネクタの最後には保護カバーを接続して下さい。

2)

アナログ出力の接続
・ 許容負荷抵抗は 500〔Ω〕以下です。
・ 極性にご注意下さい。

3)

デジタルアウトプット 1～12 の接続
・ オープンコレクタ出力です。負荷定格、極性に注意して配線して下さい。（負荷定格・ 極性にご注意下さい。
1kHz 以上の高速カウンターを使用して下さい。( 実績機種 オムロン H7CX )Vcc

DC30V、10mA）

Max 10mA+

負荷-signal－+
Max 30V

SFC011C デジタルアウトプット4)

通信の接続
・ SFC011C のパラメータを設定します。
（RS-485 通信コンバータ（別売）が必要です）
・ 極性にご注意下さい。

11

IM-F2046-J00

2. パネル部の名称と機能

2

1

3

4

6

5

7

2.1 名称と機能
表 7. パネル部説明
No.

名称機能

①

RS-485 メンテナンスポート

Modbus 通信のメンテナンス用のポートです。

②

ゼロ点調整スイッチ

ゼロ点調整を行う際に押します。

③

チャンネルセレクトスイッチ

ゼロ点調整の実施や流量、ステータス表示させるチャンネルを設定します。

④

アドレススイッチ H（×10）⑤

アドレススイッチ L（×1）

⑥表示器

⑦

チャンネルランプ

Modbus 通信のアドレス設定に使用します。

2.1.1

瞬時流量表示およびエラーメッセージを表示します。
表示器に表示されている流量、ステータスのチャンネルを示します。

ゼロ点調整スイッチZERO ADJ スイッチを 3 秒以上長押ししますとゼロ点調整が開始します。
ゼロ点調整時間は約 20 秒です。
チャンネルセレクトスイッチの設定状況でゼロ点調整を実行するチャンネルが変わります。
表 8. チャンネルセレクトスイッチ説明（ゼロ点調整）
チャンネルセレクトスイッチ
ゼロ点調整実行チャンネル
設定番号0

全チャンネル一括ゼロ点調整

1

CH1 のみゼロ点調整

2

CH2 のみゼロ点調整

3

CH3 のみゼロ点調整

4

CH4 のみゼロ点調整

5

CH5 のみゼロ点調整

6

CH6 のみゼロ点調整
12IM-F2046-J00

2.1.2

アドレススイッチ
ADRS. SWITCH の H にアドレスの 10 の位、L にアドレスの 1 の位を設定して下さい。
アドレス設定範囲は 01 ～ 32 です。
コンバータアドレススイッチと各チャンネルのスレーブアドレスの対応は以下の通りになります。
コンバータ
アドレススイッチ

XXCH1

スレーブ
アドレスXX

CH2

XX + 1CH3XX + 2CH4XX + 3

CH5XX + 4

CH6XX + 5チャンネル

設定例
SFC011C 2 台で 12 ラインをスレーブアドレス 8 から 19 で管理する場合のアドレススイッチ
設定と各チャンネルのスレーブアドレスの対応。
SFC011C No.1
アドレススイッチ

14

チャンネル

CH1スレーブア
ドレス08

CH1スレーブア
ドレス
14

CH209

CH215

CH310

CH316

CH411CH417CH512CH518

CH613CH619チャンネル

2.1.3

SFC011C No.2
アドレススイッチ

08

表示器

2.1.3.1 表示内容
表示器に表示される内容は以下の通りです。
表 9. 表示器説明
項目

流量表示

LED 表示内容説明/対応方法

XXXX

瞬時流量を 4 桁で表示します。流量単位は検出器の設定で変わります。
小口径：mL/min、大口径：L/min
ゼロ点調整中の状態を表します。

ゼロ点調整中ファームウェアダウンロード

ゼロ点調整エラー0Adj
ファームウェアダウンロード中の状態を表します。

-d1-

0-Er

ゼロ点調整に失敗しました。下記の項目を確認して下さい。
・配管内に気泡が混入していないこと。
・検出器ケーブルが正しく接続されていること。
・流量調整バルブが完全に閉になっていること。

13IM-F2046-J00

エンプティセンサー

b1Er

逆流b2Er

最大値異常

b3Er

UP/DOWN レベル差異
常b4Er

周波数異常

b5Er

減衰

EEPROM 障害

超音波信号を正しく受信できません。下記の項目を確認して下さい。
・配管内に気泡が混入していないこと。
・検出器ケーブルが正しく接続されていること。
・検出器ケーブルが断線していないこと。
流量が逆流しています。下記の項目を確認して下さい。
・検出器の向きが正しく設置されていること。
超音波信号が大きすぎます。下記の項目を確認して下さい。
・配管内に気泡が混入していないこと。
・測定流体の物性（液種、濃度）が変化していないこと。
・測定流体の液温が上昇していないこと。
超音波信号の UP 波形/DOWN 波形レベル差が大きすぎます。下記の項目を確認して下さい。
・配管内に気泡が混入していないこと。
・検出器に異常がないこと。
超音波信号の周波数が変動しています。下記の項目を確認して下さい。
・配管内に気泡が停滞していないこと。
・流量計周辺にノイズ発生源がないこと。

b6Er

超音波信号が減衰しています。下記の項目を確認して下さい。
・配管内に気泡が混入していないこと。

E-Er

EEPROM リードライト時のタイムアウト、チェックサムエラーを検出しました。
DC24V 電源の瞬断を検出しました。

電源瞬断検出

P-Er
ウォッチドッグタイマタイムアウトによる再起動を検出しました。

再起動

U-Er

※上記エラー表示は、エラー状態から復帰した後も発生したエラーを通知するために保持します。（エラー表示⇔流量表示を交互に表示します。）エラー表示をリセットする場合は、ゼロ点調整ボタンを押して離すとリセットされます。エラー表示後に再度ゼロ点調整ボタンを押すと、ゼロ点調整が開始するのでご注意下さい。

称 径
流量単位
表示範囲L/min-99.9 ～ 99.99

流量範囲[L/min]
表示 状態
100
≦ Flow
99．99 点滅
10≦ Flow ＜ 100 XX．XX 点灯
0≦ Flow ＜ 10 X．XXX 点灯-1＜ Flow ＜ 0 -．XXX 点灯-10＜ Flow ≦ -1 -X．XX 点灯-100＜ Flow ≦ -10 -XX．X 点灯
Flow ≦ -100 -99．9 点滅

14

IM-F2046-J00

2.1.3.2 表示動作
チャンネルセレクトスイッチの設定状況で表示するチャンネルが変わります。
表 10. チャンネルセレクトスイッチ説明（表示動作）
チャンネルセレクトスイッチ
設定番号
0
1
2
3
4
5
6

1 設定時

チャンネルランプ表示器

全チャンネル表示
CH1 のみ表示
CH2 のみ表示
CH3 のみ表示
CH4 のみ表示
CH5 のみ表示
CH6 のみ表示

1～6 設定時の表示動作：・表示器：
設定したチャンネルのみの表示になります。
・チャンネルランプ：”b1Er”が発生しているチャンネルが点灯します。

0 設定時の表示動作：
CH1～CH6 まで 1 秒ごとに表示が切り替わります。
また、チャンネルランプも連動します。

CH1 表示チャンネル

CH1 表示点灯

チャンネルランプ
表示器

点灯
点灯
2 設定時

1秒

CH2 表示CH2 表示
点灯

点灯
点灯
3 設定時

1秒

CH3 表示CH3 表示

点灯

点灯

点灯
4 設定時

1秒

CH4 表示CH4 表示

点灯
点灯
点灯
5 設定時

1秒
CH5 表示CH5 表示
点灯
点灯

点灯
6 設定時

1秒

CH6 表示CH6 表示

点灯
点灯

点灯

1秒
例：CH2、CH6 が b1Er、CH4 が 0-Er の場合

15

IM-F2046-J00

3. 運転本器は納入に先立ち、ご指定の仕様に基いてデータ設定・調整がされています。
取付および配線が完了した後、電源を投入すれば流量信号が得られます。
万一運転開始時に不具合が生じた場合は、PC より RS-485 の通信を使用し、設定データの確認、再設定を行って下さい。
また特にご指定のない機能については標準設定値に設定されておりますので、必要に応じて設定データの変更を行って下さい。

3.1 運転準備注意

3.1.1 電源投入前の確認事項

注記

取付および配線が完了したら、電源投入前に次の点を必ず確認して下さい。
1) 配線
・電源および出力端子の配線に誤りがなく、確実に端子に接続されていること。
・接地が確実に行われていること。
・検出器と変換器は正しい組合せで接続されていること。
2) 電源電圧が仕様と合っていること。
3) 流量の流れ方向と検出器の矢印方向が一致していること。

3.1.2 検出器通水

注記

検出器測定管路内を満水状態にして、流体を静止させて下さい。
この際、バルブにリークがなく、完全に流体が静止していることを確認して下さい。
また、気泡の発生や溜まり部がないことも確認して下さい。
気泡が溜まっている場合は、しばらくブローして、完全に気泡を除去して下さい。

3.2 運転3.2.1 通電1)2)変換器に通電して下さい。
規定の性能を満足させるため、通電後約 15 分間ウォームアップして下さい。

3.2.2 運転1)
2)

流体を流し運転を開始して下さい。
流量出力信号が、正方向の流れにもかかわらず増加しない場合は、流れ方向が逆になっている事が考えられます。検出器の取付方向と流れ方向が一致していることを再確認して下さい。

注記

3.2.3 ゼロ点の調整

配管内の気泡が抜けるまで充分に通水して下さい。
二次側バルブを閉じてから一次側バルブを閉じ、検出器内を完全に満水で静止状態にして下さい。
次にパネル面の ZERO ADJ スイッチを 3 秒以上長押してゼロ点調整を開始して下さい。
ゼロ点調整時間は約 10 秒です。終了すると自動的に測定モードに戻ります。
（ゼロ点調整中は表示器に“ 0AdJ ”と点滅表示されます）
ゼロ点調整後、出力が 0%になっている事を確認して下さい。大きくずれている場合、若しくは表示器に”0-Er”と点滅表示されている場合は、もう一度通水からやり直して下さい。
ローカットオフ機能を使用している場合、出力は設定値以下マイナス側もすべて 0%になりますのでご注意下さい。

16IM-F2046-J00

4. パラメータ
パラメータ
Detector
Full Scale
K Factor
Damping Time
Low Cut
Kinematic Viscosity
Hold Time
Burn Out

表 11. パラメータ一覧
単位
内容
検出器の口径mL/min、L/min
フルスケール流量
検出器固有の定数s
流量出力応答時間%
ローカット設定値
mm2/s
動粘度補正係数s
異常時の保持時間%
ホールド時間以上経過した場合の出力設定

5. 通信関連
5.1 パラメータ設定
パラメータ設定は SFC011C 専用コンフィグソフトで行います。

PC
RS-485
通信コンバータ
通信ケーブル

Txd+
Rxd－

USB ケーブル

SFC011C
コンフィグソフトで設定したパラメータは通信で SFC011C に送りますので上図の様に PC と SFC011C の間に RS-485 通信コンバータを接続します。
※通信コンバータは別売りですので市販の通信コンバータをご購入下さい。
動作確認機種：（株）システムサコム販売 USB-485I RJ45-T4P
パラメータ設定方法については別紙「SFC011C コンフィグソフトマニュアル」を参照して下さい。

5.2

通信仕様
別紙「SFC011C 通信仕様書」を参照して下さい。

5.3

アドレス設定☞2.1.2 アドレススイッチ参照。

17IM-F2046-J00

6. 日常点検

注記

超音波流量計は可動部や消耗部品がなく、ほとんどメンテナンスフリーでご使用いただけますが、長期に渡って安定してご使用いただくために以下の日常点検を実施することをお奨めします。

6.1 接続配管の点検
・配管に曲がりが生じて検出部に過大な応力が加わっていないか。
・配管振動は大きくないか。

6.2 防水性の点検
・検出器ケーブル接続口のシールは完全か。

6.3 検出器の点検・継手接続部および測定管内に異物または、気泡溜まりがないか、目視により確認する。

7. トラブル一覧
超音波流量計のトラブルは、配線など設置に起因するもの、測定流体に起因するもの、計器自体の故障などさまざまな原因が考えられます。
原因検索にはトラブルの現象を正確に把握し、それぞれに応じた対応をとることが近道です。
ここでは、一般的に考えられるトラブル現象別に想定される原因を記載しています。トラブル現象を確認し、ご対応下さい。
トラブルの現象想定される原因、対応方法①パラメータが最適な値に設定されているか確認して下さい。
②流れ方向が正常か確認して下さい。

1

流体を流しても出力がゼロのまま変化しない。

2

ゼロ点が不安定。

3

ゼロ点で出力が出る、振り切れる。

4

流体を流すと出力が不安定または振り切れる。 ①大きな気泡が時々混入して来ないか確認して下さい。
②パラメータが最適な値に設定されているか確認して下さい。
実流量と出力が合わない。
③配管に漏れがないか確認して下さい。

5①配管内に気泡が停滞していないか確認して下さい。
②流量調整バルブが完全に閉になっているか確認して下さい。

18

IM-F2046-J00

■ サービスネット
製品の不具合などの際は弊社営業担当か、弊社営業所までご連絡ください。
営業所については弊社ホームページをご覧ください。

■ 製品保証弊社ホームページをご覧ください。

All right Reserved Copyright © 2017 TOKYO KEISO CO., LTD.
本書からの無断の複製はかたくお断りします。