Specifications
Spherical power

Measuring range+27.00 to -27.00D
Measuring step
0.25D/1.00D/2.00D/3.00D
Measuring range+8.00 to -8.00D
Measuring step
0.25D/1.00D
Measuring range
0 to 180°
Measuring step
1°/5°/15°
Measuring range
0 to 20Δ(all direction)
Measuring step
0.1Δ/0.2Δ/0.5Δ /1.0Δ
Adjustment range
48 to 80mm
Adjustment step
0.5mm/1.0mm
Jackson Cross cylinder±0.25D/±0.50D
Auto Cross cylinder±0.25D
Red-Green filter, Polarizing filter (45°/135°), Prism (6Δ /10Δ), Red Maddox (horizontal/vertical),Lens for retinoscopy (+1.5D/+2.0D), Cross cylinder for measuring presbyopia (±0.50D), Occluding plate,Pinhole, and Cross hairs glass.

Cylinder power
Cylinder axis
Prism
Pupillary distance
Cross cylinder
Test lens (Aux lens)Reference eyeglass
wearing distance
Convergence
Forehead rest adjustment range
Dimensions and weight

12mm, 13.75mm, 16mm, 18mm, 20mm
Near-point distance 40/67cm (Minimum pupillary distance at near-point 40cm convergence : 53mm)
15mm
294-328mm (W) x 115mm (D) x 261mm (H) 4.0kg
Measuring head(Non-arm mounted type)294-328mm (W) x 115mm (D) x 324mm (H) 4.6kg
Measuring head(Arm-mounted type)
276mm (W) x 197mm (D) x 117mm (H) 2.7kg
Power Supply Unit W/External PC
276mm (W) x 197mm (D) x 117mm (H) 3.9kg
Power Supply Unit W/Built-in PC
300mm (W) x 250mm (D) (MAX) x 200mm (H) (MAX) 2.3kg
1Dial Controller

System Chart
Ez MeterComputerized
Lensmeter
1 Dial Controller

KB-50S

Auto Refractometer
Auto KeratoRefractometer

Power Supply Unit
w/Built in PC
TabletMirror Chart
Compu Vision

Mouse & Monitor

CV-5000PRO

Power Supply Unitw/External PCPC

Mouse & Monitor

Auto Chart
Projector

Wave-Front
Analyzer

LCD Chart(wire or wireless)* Not available in all countries, please check with your distributor for availability in your country* Subject to change in design and/or specifications without advanced notice.

IMPORTANTIn order to obtain the best results with this instrument,
please be sure to review all user instructions prior to operation.

75-1 Hasunuma-cho, Itabashi-ku, Tokyo 174-8580, Japan. Phone:+81-(0)3-3558-2522/2502 Fax:+81-(0)3-3965-6898 www.topcon.co.jp

TOPCON MEDICAL SYSTEMS, INC.

TOPCON S.A.R.L.

TOPCON CANADA INC.

TOPCON DEUTSCHLAND MEDICAL G.m.b.H.

TOPCON EUROPE MEDICAL B.V.

TOPCON SCANDINAVIA A.B.

111 Bauer Drive, Oakland, NJ 07436, U.S.A.
Phone:+1-201-599-5100 Fax:+1-201-599-5250 www.topconmedical.com
110 Provencher Avenue, Boisbriand, QC J7G 1N1 CANADA
Phone:+1-450-430-7771 Fax:+1-450-430-6457 www.topcon.ca
Essebaan 11; 2908 LJ Capelle a/d IJssel; P.O.Box145;
2900 AC Capelle a/d IJssel; THE NETHERLANDS
Phone:+31 -(0)10-4585077 Fax:+31 -(0)10-4585045
E-mail: medical@topcon.eu; www.topcon-medical.eu

ITALY OFFICE

Viale dell' Industria 60; 20037 Paderno Dugnano; (Milano), ITALY
Phone:+39-02-9186671 Fax:+39-02-91081091 E-mail: info@topcon.it; www.topcon.it

BAT A1 3 route de la révolte 93206 SAINT DENIS CEDEX
Phone:+33-1-49-21-23-23 Fax : +33-1-49-21-23-24 E-mail:topcon@topcon.fr; www.topcon.fr
Hanns-Martin-Schleyer Strasse 41; D-47877 Willich, GERMANY
Phone:+49-(0)2154-8850 Fax:+49-(0)2154-885177 E-mail:med@topcon.de; www.topcon.de
Neongatan 2; P.O.Box 25; 43151 Mölndal, SWEDEN
Phone:+46-(0)31-7109200 Fax:+46-(0)31-7109249 E-mail:medical@topcon.se; www.topcon.se

TOPCON ESPANA S.A.
HEAD OFFICE

Frederic Mompou 4 Esc. A Bajos 3, 08960 Sant Just Desvern Barcelona, SPAIN
Phone:+34-93-4734057 Fax:+34-93-4733932 E-mail: medica@topcon.es; www.topcon.es

DANMARK OFFICETOPCON ( GREAT BRITAIN ) LTD.

IRELAND OFFICETOPCON POLSKA Sp. z. o. o.

Praestemarksvej 25; 4000 Roskilde, DANMARK
Phone:+45-46-327500 Fax:+45-46-327555
E-mail: topcon@topcondanmark.dk www.topcondanmark.dk
Unit 276, Blanchardstown; Corporate Park 2 Ballycoolin Dublin 15, IRELAND
Phone:+353-18975900 Fax:+353-18293915 E-mail: medical@topcon.ie; www.topcon.ie

Topcon House,Kennet Side,Bone Lane,Newbury,Berkshire RG14 5PX United Kingdom
Phone:+44-(0)1635-551120 Fax:+44-(0)1635-551170 E-mail:medical@topcon.co.uk; www.topcon.co.uk

ul. Warszawska 23; 42-470 Siewierz, POLAND
Phone:+48-(0)32-6705045 Fax:+48-(0)32-6713405 www.topcon-polska.pl

TOPCON SINGAPORE MEDICAL PTE. LTD.

1 JALAN KILANG TIMOR #09-01 PACIFIC TECH CENTRE SINGAPORE 159303
Phone:+65-68720606 Fax:+65-67736150
E-mail:medical_sales@topcon.com.sg www.topcon.com.sg

REPRESENTATIVE OFFICE IN INDONESIA

Level 38, Tower A, Kota Kasablanka unit GH-04, JI.
Casablanca, Kav 88, Jakarta, Indonesia 12870
Phone:+62-21-2963-8004

TOPCON INSTRUMENTS ( MALAYSIA ) SDN.BHD.
No. D1, (Ground Floor), Jalan Excella 2, Off Jalan Ampang Putra,
Taman Ampang Hilir, 55100 Kuala Lumpur,MALAYSIA
Phone:+60-(0)3-42709866 Fax:+60-(0)3-42709766

TOPCON INSTRUMENTS ( THAILAND ) CO.,LTD.
77/162 Sinnsathorn Tower, 37th Floor,Krungthonburi Rd.,Klongtonsai,
Klongsarn, Bangkok 10600,THAILAND
Phone:+66(0)2-440-1152~7 Fax:+66-(0)2-440-1158

Mehra Eyetech Private Limited

801 B Wing, Lotus Corporate Park, Graham Firth Steel Compound Goregaon (East)
Mumbai 400063 Maharashtra, INDIA
Phone:+91-22-61285455 www.mehraeyetech.in

Printed in Japan 2016_12_30_NP_CV5000_E149-2_TEM_TSM

CV-5000 PRO
Compu VisionComfort and Efficiency
in a Compact System
CV-5000PRO allows for comfort and efficiency in your refraction. Easy understandable layouts andvarious programs to support a smooth operation, offers free stress refractions for operators andpatients. A quick operation with various controllers and wide connectivity with other devices supports

CV-5000PRO Feature
Topcon’s Smart Cross System
Topcon’s original system, Smart Cross offers asmooth operation for cylinder test. Seeing thedifferent values on flapping, a patient is enable topick up easily which one they can better distinguish.

efficient workflow. Comfort and Efficient operations creates better communication with patients, formore satisfaction of refraction results.

Color display for cross cylinder
CV-5000PRO displays the chart in a red or green
circle, during cross cylinder tests. A patient quickly
mentions their choice, and it enables accurate
refraction results.

Compact, Open andContemporary Design
The open and compact design allows perfectinteraction with your patient.

The operator isenables to monitor the patient’s expression
during the refraction.

High Speed Lens Disc Rotation
Efficient lens disc rotation reduces ocular stress aswell as confusion for patients. The total refraction
time is reduced significantly.

Range of control solutions
to suit your practiceCV-5000PRO is available with a range of control options, allowing you to tailor the solution tomeet your needs and optimize workflow in your practice.

Control Options
KB-50SThe KB-50S is a dedicated controller for theCV-5000PRO vision tester. This controller enables
easy and intuitive operations of the CV-5000PRO. For
quickandeasy

operation,

thebuttons

areergonomically placed around a central dial control. A
large 10.4” color LCD touch screen monitor also
provides a range of options.

Tablet*
The wireless tablet option offers a flexible operation
within the refraction room; the system can be operated
from any location in the room, and control of therefraction system can be combined with other usefulclinical tablet applications to optimize patient care. The
tablet enables intuitive operation through the touchingand swiping of the screen. An operator enables
position any location during the refraction, and also
use other application software on the same tablet forbetter consulting.

Operating software on theintegrated PC*
Instead of the KB-50S as a controller, software on
an integrated CV-5000PRO system PC allows
control simply by connecting a mouse and LCD
monitor. The user interface normally displayed on aKB-50S is viewed on the LCD monitor.

Operating software on external PC*
A unique feature in this market is the ability to control
the CV-5000PRO vision tester through your own
Personal Computer, enabling you to use othersoftware simultaneously.

* Not available in all countries, please check with your distributor for availability in your country.
*Please check with your distributor about abailable type of tablet and PC.

Easy to Use

Examination Support Features

Intuitive Layout

Personalized Chart Pages

The newly redesigned interface is

A wide range of test chart options are laid out on fiveflexible and easy to use.

tab pages, accessed via the touch panel. This includes

All relevant data can be viewed at athe conventional charts used by the Topcon chart

glance. The layout can be modified

projectors. The layout of the pages can be defined by

to suit the users needs.

the user, allowing rapid access to favoured charts.

Multiple "HELP" Function
At the touch of a button the HELP display is

Dial Navigation Function (KB-50S)

activated. HELP provides useful information on

The user can select to use the dial navigation function,

the selected test. A simplified (short text) or anallowing precise control over the sphere, cylinder or prism

advanced (longer text) HELP display can be

power in response to the patient's answers. The ultimate inselected.

comfort and ease of use.

Pre defined examinations
ExampleThe KB-50S provides pre defined examinations to correct
the spherical, cylindrical refraction as well as phoria
tests and binocular function teste.

Other SpecificationData Import & Export Functions

Range of illustrations for better communication with patients

Easy Operation by networking

CV-5000PRO provides a range of illustrations to help explain progressive lenses, accommodation, and refractive status.

CV-5000PRO is able to receive refraction data

Communication with easy understandable animation supports patient’s understanding of the examination results.

from

other

Topconinstruments,

such

as

refractometers and lensmeters, so that an

Myopia /HyopiaProgressive
Lens

KR-1

operator is able to log this data rapidly on to thecontrol monitor at the start of an examination.
Automatic

datatransfer

avoidsmistakes

associated with manual typing and makes theroutine more efficient. An efficient workflow has
benefits both for your patients and your practice.

PCCL-300
CV-5000PRO
CV-

Far-NearAccommodation
KR-800Age 10Age 50Integration in IMAGEnet series/
other software

Tilting Function for Near Vision Tests

The CV-500PRO software integrates with theTo provide a natural reading position during the nearlensmeter data can be transferred from thetests, the CV-5000PRO head can be tilted. The tilting

CV-5000PRO system to the IMAGEnet system andangle can be fixed with in a wide range.

assigned to a patient. The CV-5000PRO enables

IMAGEnet system*. Subjective, objective andalso the export of data to other software within thepractice.
* IMAGEnet Software is optional

Near vision LED Illumination
LED illumination from the CV-5000PRO vision
tester provides bright illumination of the near test
chart. The use of external illumination is no longer
required.

Near vision LED

More choice for your system

KR-800 series

MC-4S Mirror Chart

Auto Kerato-Refractometer(KR-800 / KR-800A /KR-800S)

Space-saving
Liquid crystal polarization

Auto Kerato-Refractometer with objective andsubjective testing (KR-800S)

High resolution
3D Contents
Smooth Operation

3D Auto Alignment function allows quicker andeasier measurement (KR-800A)

Active Charts

Smooth operation
8.5 inch wide touchscreen control panel
Compact designed body
Easy set-up with a LAN network

PC-50S/ PC-50SB Pixel Chart
Comprehensive Range of Optotypes
True Visual Acutity Testing System
The Variety of Chart, Image and Movie
Mimimal Design

KR-1

Auto Kerato-Refractometer

Active Charts
Simple and Quick Operation
Easy-to-Use Color Touchscreen Control Panel
Flexible and Space Saving Layout

ACP-8 Auto Cahrt Projector

New Compact and Ergonomic Design
Reliable Measurement

Easy to Use
Easy Setup and Maintenance
Program Settings
Connectivity with CV-500PRO
Tilt function

TRK-2PAuto Kerato-Refracto Tonometer

CL-300 Computerized Lensmeter

Refractometer, Keratometer, Non-Contact
Tonometer and Pachometer in one single instrument

LCD Touch Panel

Fully automatedCompact Slim Body

Rotating touch screen control panel

Automatic Mono- & Multi-focal Detection

Compact and modern design

UV Transmittance Measurement

Easy manual alignment withnew control lever accessory

Green Measurement Light Beam

LU-1

Control lever* Not available in all countries, please check with your distributor for availability in your country.