Digital OscilloscopeDS-8000 Series

350MHz/500MHz/1GHz
5GS/s, 8CH/4CH Made in Japan

High Cost-performance

The New Standard of High Resolution Digital Oscilloscope■ Brilliant Display■ Intuitive Operation■ Fast Boot-up
Large 15.6-inch FULL HD with Touch Screen, Buttons and Knobs Approx. 30 seconds

Max. 8 Channels
High Resolution 12-bit(High Resolution Mode 16-bit equivalent)Max. Memory Length 120M-point as standard
High-speed Data Transfer(15 times faster than conventional models)Large Size Screen・Intuitive Operability■ Brilliant Dispaly■ Intuitive Operation■ Fast Boot-up

Easy-to-read Measurement Parameter
Display. Large Size Full HD Monitor
displays waveform details.

Comfortable operation with user-friendly
Touch Screen, Buttons and Knobs.

Quick Start-up can increase efficiency
in measurement operation (Approx. 30
seconds after Turning-ON the power).

①⑤④③⑥②Front Panel

Left Side Panel① 15.6-inch Large Size Touch Screen⑦ USB/LAN Interface
for Remote Control② Input Channels 8CH/4CH③ Probe Power Supply
for IWATSU Branded Current/High Voltage Differential Probes④ Measurement Parameter Display
Max. / Min. / Ave. values for up to 8 parameters⑤ ZOOM Function
Two ZOOM buttons allow to enlarge waveforms in different
positions.
⑥ USB Host Terminal（Front 2, Left Side 1）
Mouse and keyboard can be connected.
Display Image and waveform data can be saved to USB
memory.

⑧ DVI-D Interface
The screen can be output to an
external display.
⑨ AUX OUT Terminal⑥⑦⑧⑨

Measurement Example: Switching voltage / current of GaN Device evaluation board
Rogowski Coil Current Probe (IWATSU)

High Voltage Differential Probe
BumbleBee (PMK)Uncompromised Basic performance■ Full-Time High Resolution 12-bit & Low Noise, Max. 16-bit equivalent (High Resolution Mode)
12-bit ADC

16-bit waveform display with the High Resolution Mode
●Application:
For switching waveforms (ON voltage, etc.) of a power device, lightning surge
waveform observation etc.
It can be used in R&D, production and QA of automotive components andSemiconductors etc.

50V/div

VDSON Voltage
ICComparison of Vertical Resolution

VDS

5V/div

ON Voltage

5GS/s

Enlarged waveform of Power device (GaN) switching waveform and ON voltage (VDS)【 8-bit】【16-bit】

Large noises are observed when enlarged with
8-bit due to effect of low resolution.

16-bit equivalent resolution shows more details ofenlarged waveform.

■ Analog Channel Max. 8CH■ Simultaneous enlargement of 2 different positions

Frequency Bandwidth : 1GHz/500MHz/350MHz
Max. Sampling Rate : Max. 5GS/s (Channel Combine)
Max. 2.5GS/s (All CH)
Max. 8CH Waveform Operation (MATH)●Application:
For observation of voltage/current/sensing/ECU signal waveforms ofautomotive inverters, etc.
It can be used in R&D, production and QA of automobile, machine tools,robots, and production equipment.

VDS 280V

Double Pulse

MATH=VDS × ICVDS40V/div
5µs/div
MATHICTURN ON Enlarged
50ns/div

ICICTURN OFF Enlarged

VDS

5GS/s

MATH=VDS × IC

Display of enlarged Double Pulse waveform of power device (GaN) Turn-ON and Turn-OFF waveforms.

Since the waveforms of two different positions can be enlarged simultaneously,
the Turn-ON and Turn-OFF losses of a power device can be observed.
●Application:
For loss analysis (Double Pulse Test) of Power Devices.

■ Memory Length 120M-point as standard
5GS/s
Screen dispaly example with 8CH input

2ms/div■ High-speed Data TransferEnlarged
100ns/div

Achieved 15 times faster transfer rate than conventional models.
High-speed transfer of High resolution waveform, and measurement parameters.
● Improvement of Takt Time of production
equipment● Suitable for high-speed transfer ofwaveform data for simulation

5GS/s

Before inspection
Inspection
After Inspection**A**V

Memory length up to 120M-point (Channels combine), 60M-point (All channels)
The Replay function (memory sprit display) allows to identify changes in the waveform after
measurement.
●Application:
For observation of Behavior of equipment after Turn-ON, Stability of aPower Converter, and Changes in waveform distortion of encoder pulses.
It can be used in R&D, production and QA of automobile, machine tools,robots, and production equipment.
Also suitable for Loss analysis (Double Pulse Test) of Power Devices.

■ Compatible with variety of probes
DS-8000 Series can be connected to various Current and Voltage probes.

Rogowski Coil Current Probe

High Voltage 400MHz Deferential Probe

High Voltage Probes

FET Probes■ SS-280A-H Sereis■ BumbleBee■ PHV Sereis■ TETRIS Sereis-40ºC to 150ºC , 30A to 12kApeak

Max. 2kVpeak (at 500:1)

Max. 4kV peak

Frequency bandwidth Max. 2.5GHz

DS-8000 Sereis Specifications
Specifications
Model
Input Channels
Frequency Bandwidth

DS-8000 Sereis

DS-8034

DS-8038

DS-8054

DS-8058

DS-8104

DS-8108

484848350MHz

500MHzInput Impedance

50Ω±1.5% / 1MΩ±1%//16pF±2pF

Input Coupling

GND, DC1MΩ, AC1MΩ, 50Ω

Input Sensitivity

1mV to 1V/div (50Ω, 1-2-5 Sequence)
1mV to 10V/div (1MΩ, 1-2-5 Sequence)

ADC Resolution

Real Time: 12-bit, High Resolution Mode: Max. 16-bit

Time base Range

500ps/div to 50s/divTime base Accuracy

200ps/div to 50s/div
less than 10ppm

Acquisition memory length

60M-point (All channels) / 120M-point (Channel Combine)
5GS/s @12-bit (Channel Combine)
2.5GS/s @12-bit (All channels)

Sampling Rate
Trigger

1GHz

Edge, Edge OR, Edge Alternate, Pulse Width, Period, Pattern, Dropout
Serial Trigger (UART, I2C, SPI),Sequence Trigger

Trigger Coupling

AC, DC, HF Rej, LF Rej, Noise Rej

Operation (MATH)

Addition, Subtraction, Multiplication, Division, Integration, Differentiation, Absolute Value, Inversion, FFT

Parameter Measurement

Frequency CounterVertical Axis (12 items):
Maximum, Minimum, Peak-Peak, RMS, Cycle RMS, Mean, Cycle Mean, Top, Base, Top-Base,+Overshoot, -OvershootHorizontal Axis (14 items):
Transition Time, Trise(20-80%), Tfall(80-20%), Trise(10-90%), Tfall(90-10%), Trise(Level), Tfall(Level), Frequency, Period,+Pulse Count, -Pulse Count, +Pulse Width, -Pulse Width, Duty Cycle
Others (9 items):
dV/dt, Integral, Integral(Absolute), Integral(Positive), Integral(Negative), Skew(%), Skew(Level), Phase(%), Phase(Level)
Resolution: 6-digit, Frequency Bandwidth: 4Hz to 1GHz (Upper limit is limited by the frequency bandwidth of the model)
Accuracy: 10ppm

Display

15.6-inch, Full-HD (1920x1080), Electrostatic Touch Screen

Remote Interface

LAN, USB

Power Supply

100 to 240V AC, 50 to 60Hz

Demensions

445 (W) x 324.6 (H) x 200 (D) mm

WeightApprox. 11.5kg

Power Consumption

360VA (325W)

Standard Accessory

Panel Cover, Power Cord, Passive Probe x 4, Instruction Manual (CD), Accessory Storage Bag※The products shown in this catalogue are current models at the date of publication. Design and specification are subject to change without notice for improvement.
※All enterprises including National instruments and Microsoft, etc. and product names mentioned are trademarks or registered trademarks of the respective owners.
※Some of the products are Regulated Products subject to the Foreign Exchange and Foreign Trade Control Law of Japan. Export should not be allowed without appropriate governmental authorization. Please ask our sales office whether the product concerned is a Regulated Product(s).

https://www.iwatsu.com/tme

Overseas Sales Sect.
Test & Measurement Solution Sales Dept.
7-41, Kugayama 1-chome, Suginami-ku, Tokyo, 168-8501 Japan
Tel: +81-3-5370-5483 Fax: +81-3-5370-5492

C.S(OG)2307-857-3-00