フィルターパックカタログ

Teflon® フィルターパック
25mmフィルターパック
パーソナルサンプリング
URG-2000-25FG

SPECIFICATIONS:• Designed to Contain 25mm Filter Media• Quick Release at Inlet for Attaching to Personal Sampling Cyclone• Includes 25mm Stainless Steel Support Screen• 1 3/8” Largest Outer Diameter• 3 1/4” Height• 0.036kg

37mm Teflon® フィルターパック
URG-2000-22F/22FB

URG-2000-22FB

SPECIFICATIONS:• Designed to Contain 37mm Filter Media• #30 Thread at Inlet to Allow Direct Connection
to Annular Denuders, Manifolds, Cyclones and OtherURG Components• Filters Separated by Teflon® Spacers• 2 1/4” Largest Outer Diameter• 4 1/4” Height (URG-2000-22F, 2-Stage)• 3 1/2” Height (URG-2000-22FB, 2-Stage)• 0.24kg (URG-2000-22F)• 0.2kg (URG-2000-22FB)

URG-2000-22F

Teflon® フィルターパック
47mm Teflon® フィルターパック
URG-2000-30F/30FG/30RAF シリーズ
SPECIFICATIONS:• Designed to Contain 47mm Filter Media• #30 Thread at Inlet to Allow Direct Connection
to Annular Denuders, Manifolds, Cyclones andOther URG Components• Filters Separated by Teflon® Spacers• 2 5/8” Largest Outer Diameter• 5” Height (2-Stage)• 0.45kg
FEATURES:• Stainless Steel Support Screen is Teflon®
Coated to Prevent Reactions Between Screenand Filter Media• Design Ensures Sample Integrity and Allows forEasy Transportation to and from Lab• Uses Novel O-Ring Arrangement to Prevent
Contact Between O-Ring and Filter Media andCreate a Leak-Proof Seal
• Available with 1 to 4 Stages to Allow for Sampling on Multiple Filters at Once
*See Price List For Individual Part Numbers• Quick Release Fitting at Outlet for Easy attachment to Vacuum Pump

URG-2000-30F Series

URG-2000-30FG Series

URG-2000-30RAF Series

Teflon® フィルターパック
47mm Teflon® フィルターパック
URG-2000-30FH/30FB シリーズ
SPECIFICATIONS:• Designed to Contain 47mm Filter Media• #30 Thread at Inlet to Allow Direct Connection toAnnular Denuders, Manifolds, Cyclones and OtherURG Components• Filters Separated by Teflon® Spacers• 2 5/8” Largest Outer Diameter• 4 1/4” Height (2-Stage)• 0.23kg
FEATURES:• Stainless Steel Support Screen is Teflon® Coatedto Prevent Reactions Between Screen and Filter Media• Design Ensures Sample Integrity and Allows for Easy
Transportation to and from Lab• #30 Female Thread at Outlet to Attach URG Components
such as PUF Sampler• Uses Novel O-Ring Arrangement to Prevent ContactBetween O-Ring and Filter Media and Create aLeak-Proof Seal
• Available with 1 to 4 Stages to Allow for Sampling on
Multiple Filters at Once
*See Price List For Individual Part Numbers

URG-2000-30FH Series

URG-2000-30RAF Series

90mm フィルターパック
URG-2000-30FK/30FJ シリーズ
SPECIFICATIONS:• Designed to Contain 90mm Filter Media• Filter Separated by Teflon® Coated Spacers• 4 1/2” Largest Outer Diameter• 5 5/8” Height (URG-2000-30FJ, 2-Stage)• 3 5/8” Height (URG-2000-30FK, 2-Stage)• 1.13kg (URG-2000-30FJ, 2-Stage)• 1kg(URG-2000-30FK, 2-Stage)
FEATURES:• Stainless Steel Support Screen is Teflon®
Coated to Prevent Reactions Between
Screen and Filter Media• Design Ensures Sample Integrity and Allowsfor Easy Transportation to and from Lab• Available with #30 Thread or 1” FNPT at Outlet*See Price List For Individual Part Numbers• URG-2000-30FJ with #30 Female Thread at
Inlet to Allow Direct Connection to Annular
Denuders, Manifolds, Cyclones and OtherURG Components• URG-2000-FK with “Open Face” Inlet for Collection of Total Particles• Uses Novel O-Ring Arrangement to Prevent Contact Between O-Ringand Filter Media and Creates a Leak-Proof Seal
• Available with 1 to 4 Stages to Allow for Sampling on Multiple Filters at Once
*See Price List For Individual Part Numbers

URG-2000-30FK
SeriesURG-2000-30RAF
SeriesURG-2000-30FJ Series

47mm ステンレス フィルターホルダー
SPECIFICATIONS:• Designed to Contain 47mm Filter Media• Stainless Steel• 1/2” FNPT at Inlet and Outlet• 3” Largest Outer Diameter• 7 1/16” Height (URG-2000-30FV-QCM)• 8” Height (URG-2000-30FVT-QCM)• 1.29kg(URG-2000-30FV-QCM)• 1.68kg(URG-2000-30FVT-QCM)
FEATURES:• Can Be Used with Stainless Steel Cyclone forDiesel Emissions• Designed to Meet USEPA’s Specifications for
CFR Part 86 Subpart N Revisions for theCertification of Heavy Duty Diesel Engines• Available as Single or Dual Stage• Filter Material Contained in 47mm Delrin Filter
Cassette Specified by New USEPA Requirements• Top is Hinged to Allow for Easy Filter Cassette
Change Out

URG-2000-30RAF Series

URG-2000-30FK Series

URG-2000-30FJ Series