6-Pin TO Sensor with
Integrated Signal
Conditioning
Datasheet HEIMANN Sensor Integrated Module TO-Case
Type HIS-Lx-FL5.5-Tyyy

General information about HIS models are described in“General Datasheet HIS” .

Liability
Changes or modifications at the product which haven't influence to theperformance and/or quality of the device haven't to be announced to thecustomers in advance. Customers are requested to consult with Heimann
Sensor representatives before the use of Heimann Sensor products in special
applications where failure or abnormal operation may directly affect humanlives or cause physical injury or property damage. The company or theirrepresentatives will not be responsible for damage arising from such use
without prior approval.

Features
Small size of standard TO39 package with 6 pins
5V version available
2 analog outputs for object and sensor temperature
Low response time of 5ms
Lens optics with 8:1 D:S ratio
Absorber size 0.61mm x 0.61mm
Lens optics focal length 5.5mm
Adjustable object temperature range upto 300°C (calibration preset -> Tyyy)
IR transmission 52 % within wavelength range 5.5µm to 13.5µm
Internal compensated thermopile signal (type HIS-LC) available (-> generaldatasheet)

Ordering Information
HISLx-> Heimann thermopile sensor and analogue ASIC in TO39 housing-> „L“ cap TO39 for lens FL5.5; „U“ sensor signal not compensated ;„C“ sensor signal internal compensatedFL5.5 -> lens type – silicon uncoated , focal length 5.5mm
Tyyy -> object temperature upper limit

HEIMANN Sensor GmbHGrenzstr. 22D-01109 Dresden
Rohrbergstr. 7D-65343 EltvilleManaging DirectorDr. J. Schieferdecker
Reg. at District Court
Dresden HRB20692
VAT-ID DE813444739

Internet: www.heimannsensor.com
Mail: info@heimannsensor.com
Phone 49 (0) 6123 60 50 30
Fax49 (0) 6123 60 50 39

Datasheet HIS-Lx-FL5.5-Tyyy R02 Sep 2010 Page 1 of 4••••••••••

6-Pin TO Sensor with
Integrated Signal
Conditioning
Datasheet HEIMANN Sensor Integrated Module TO-Case
Type HIS-Lx-FL5.5-Tyyy
Field of View – Definition and Sample Measurement

parameterlimitsMin

Typunitconditions

Max0degreeSensor view direction

Field of View6degree

50% energy points

D:S Ratio

8:1

Distance to spot size

Managing Director
Dr. J. Schieferdecker
Reg. at District Court
Dresden HRB20692
VAT-ID DE813444739

Internet: www.heimannsensor.com
Mail: info@heimannsensor.com
Phone 49 (0) 6123 60 50 30Fax
49 (0) 6123 60 50 39

HEIMANN Sensor GmbH
Grenzstr. 22D-01109 Dresden
Rohrbergstr. 7D-65343 Eltville

Datasheet HIS-Lx-FL5.5-Tyyy R02 Sep 2010 Page 2 of 4

Optical axis

6-Pin TO Sensor with
Integrated Signal
Conditioning
Datasheet HEIMANN Sensor Integrated Module TO-Case
Type HIS-Lx-FL5.5-Tyyy
Specifications
Parameter

Typical Value

UnitCondition

Supply voltage VDD

4.5 .. 5 .. 5.5

V+Vs

Supply voltage VSS

0V-Vs , Ground

Supply current

1 .. 1.7 .. 2.2

mA

Without load

Output voltage range

0.3 .. VDD-0.3

VStart up time after POR

Max. 0.5

secElectrical start up

Sensor absorbing area

0.6 x 0.6mm²Type TP1

Sensor amplification

150 .. 5500

Response time sensor

5msec

t/T = 63%

Temperature reference
voltage at 25°C

1.225

V

Output AORSensitivity temperaturereference

15(10..16)mV/°C

Linear ; output AOR ; type “U”(type “C” )Operating temperature-20.. 120°C

Output AOT , preadjusted

Pin No.

SymbolDescription

1VDD

Positive supply voltage (+5V)

2

VSS

Negative supply voltage / Ground (0V)

3

AOT

Amplified analog sensor output voltage

4

AORAnalog temperature reference output voltage

5

SCLK

Adjustment mode only – serial clock input

6CM

Adjustment mode only – programming mode selection

HEIMANN Sensor GmbHGrenzstr. 22D-01109 Dresden
Rohrbergstr. 7D-65343 EltvilleManaging Director
Dr. J. Schieferdecker
Reg. at District Court
Dresden HRB20692
VAT-ID DE813444739

Internet: www.heimannsensor.com
Mail: info@heimannsensor.com
Phone 49 (0) 6123 60 50 30
Fax
49 (0) 6123 60 50 39

Datasheet HIS-Lx-FL5.5-Tyyy R02 Sep 2010 Page 3 of 4

Pin / Device Configuration

6-Pin TO Sensor with
Integrated Signal
Conditioning
Datasheet HEIMANN Sensor Integrated Module TO-Case
Type HIS-Lx-FL5.5-Tyyy
Dimensions / Pin Assignment

Managing Director
Dr. J. Schieferdecker
Reg. at District CourtDresden HRB20692
VAT-ID DE813444739

Internet: www.heimannsensor.com
Mail: info@heimannsensor.com
Phone 49 (0) 6123 60 50 30
Fax49 (0) 6123 60 50 39

Datasheet HIS-Lx-FL5.5-Tyyy R02 Sep 2010 Page 4 of 4

HEIMANN Sensor GmbHGrenzstr. 22D-01109 DresdenRohrbergstr. 7D-65343 Eltville