Software IRBIS® 3Drilling process

ImageIR® 4300

High-end Thermography with an Entry-level ModelDetector Format
Large detector enables
highest sensivity
IR-Frame Rate
Analysis of extreme temperaturechanges and gradients in full frame
Measurement Accuracy
Highly accurate andrepeatable measurements
Thermal Resolution
Precise detection of smallest
temperature differences
Shortest Integration Time
Accurate temperature measurements
of fast processes

The entry-level model ImageIR® 4300 already shows, which qualities are characteristic for the high-end camera series ImageIR®
are. Equipped with a cooled focal-plane array photon detector
with (320 × 256) IR pixels this camera enables users to choose between detectors made of different material for thermal analyses
in the short-wave and mid-infrared spectral range. MCT detectors
support snapshot mode.
Recording and storing images with frequencies up to 706 Hz enables you to analyse even fast processes. In addition, the ImageIR® 4300 comes with an impressive thermal resolution up to 0.02 K(20 mK) due to its pixel pitch of 30 µm. In sum, this camera series
provides a potential that qualifies for usage for a broad range ofapplications in the fields of industry and science.
The robust light-metal housing of the instruments matches thisclaim. With the combination of the modular designed camera
concept, the internal trigger interface, most diverse thermographic software and high-quality lenses users benefit from ahigh level of flexibility. That allows to adapt the cameras to almost
every measurement and testing task.

Pitch Dimension
Precise measurement of low temperaturesand very fast integration times
GigE Vision Compatible
Standard interface for easy integration
into existing process environment
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range(3.7 … 4.8) µm

Pitch

30 µm

Detector

MCT

Detector format (IR pixels)(320 × 256)

Image acquisition

SnapshotReadout mode

ITR

Aperture ratio

f/2.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 300) °C*, up to 3,000 °C*

Measurement accuracy± 2 °C or ± 2 %Temperature resolution @ 30 °C

Better than 0.02 K

Frame rate (full / half / sub frame)*

Up to 75 / 265 / 706 Hz

Window mode

Yes* (full frame / sub frame)

Focus

Manual, motorised or automatically*

Dynamic range

14 bitIntegration time

(1 … 20,000) µs

Rotating filter wheel*

Up to 5 positions

Rotating aperture wheel*

Up to 5 positions

Interfaces

GigE, HDMI*

Trigger

1 IN / 1 OUT, TTL

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5

Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight(241 × 120 × 160) mm*; 3.3 kg (without lens)Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,  IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on modelLenses

Focal length (mm)

FOV (°)IFOV (mrad)

Wide-angle lens12(43.6 × 35.5)

2.5Standard lens25

(21.7 × 17.5)

1.2

Telephoto lens

50

(11.0 × 8.8)

0.6

Telephoto lens

100

(5.5 × 4.4)

0.3

Telephoto lens

200

(2.7 × 2.2)

0.15

Macro and microscopic lenses

Minimum object distance (mm)Object size (mm)Pixel size (µm)

Close-up for telephoto lens 50 mm

300(58 × 46)

180

Close-up for telephoto lens 100 mm

500(48 × 38)

150

Microscopic lens M = 1.0× (2 versions)

195 / 300

(9.6 × 7.7)30

22(3.2 × 2.6)

10Microscopic lens M = 3.0×© InfraTec 04 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.