Smart way to a healthy life0123

https://www.aandd.jp

Digital Blood Pressure Monitor

Precision Health Scale

UA-651BLE

UC-352BLE

ISO plug compatible
Clinically validated for accuracy (ESH2010)
Bluetooth Low Energy (Ver.5.1)
Bluetooth communication
Internal memory of 30 data records
Irregular HeartBeat (IHB) indicator - LCD only
Extra large 3-line display for easier reading of results
One-touch measurement
Latex-free and slimFit Cuff
Compact design for portability

Bluetooth Low Energy (Ver.5.1)
Bluetooth communication
Durable glass plate scale
kg/lb selection
Internal memory of 90 data records0 kg to 100 kg : 100 g resolution / 100 kg to 200 kg : 200 g resolution
0 lb to 220.0 lb : 0.2 lb resolution / 220.0 lb to 450.0 lb : 0.5 lb resolution
Simply step on start weighing
Beeps on completion of weighing
Low profile stylish design

Specifications
UA-651BLE

UC-352BLE

Measurement method
Display typeMeasurement range

Oscillometric measurement

Measurement accuracy

Pressure : ±3 mmHg
Pulse : ±5 %Power supply

4 x 1.5 V batteries (LR6, R6P or AA) orAC adapter (TB-233) (Not included)

Number of measurements

Approx. 700 times LR6 (alkaline batteries)
Approx. 200 times R6P (manganese batteries)
With pressure value of 180 mmHg at room
temperature of 23 °C

Memory storage
Cuff
Operating conditions
Transport/Storage conditions
Wireless communication
Dimensions
WeightOptions

Digital LCD, Pressure/Pulse displayed simultaneously
Pressure : 0 mmHg - 299 mmHg
Systolic pressure : 60 mmHg - 279 mmHg
Diastolic pressure : 40 mmHg - 200 mmHg
Pulse : 40 - 180 beats / minute30
SlimFit Cuff with metal D-ring / Arm circumference : 22 cm-32 cm+10 °C to +40 °C / 15 %RH to 85 %RH/800 hPa to1060 hPa-20 °C to +60 °C / 10 %RH to 95 %RH
Bluetooth Low Energy Ver.5.1
Approx. 96 [W] x 68 [H] x 130 [D] mmApprox. 250 g, excluding batteries
SlimFit Large Adult Cuff(31 cm-45 cm) : CUF-D-LA-ISO
SlimFit Semi-Large Cuff(23 cm-37 cm) : CUF-D-MA-ISO
Wide range Cuff(22 cm-42 cm) : CUF-I-ISO
SlimFit Adult Cuff(22 cm-32 cm)：CUF-F-A-ISO
AC Adapter (C/BF plug) : TB-233C/BF

Maximum capacity
Display type
Minimum display

200 kg / 450 lb

Setting items
Power source
Battery life

Units : kg or lb

Memory storage
Operating conditions
Transport/Storage conditions
Wireless communication
Communication specifications
Dimensions
Weight90

Digital LCD, Weight displayed after step off
0.1 kg (0 kg to 100 kg) / 0.2 kg (100 kg to 200 kg)
0.2 lb (0 lb to 220 lb) / 0.5 lb (220 lb to 450 lb)
4 x 1.5 V alkaline batteries (LR03 or AAA)Approximately 1 year
(two measurements per day when alkaline batteries are used)+10 °C to +40 °C / 30 %RH to 85 %RH-10 °C to +60 °C / 30 %RH to 85 %RH
Bluetooth Low Energy Ver.5.1*Continua Design Guideline 2014
Approx. 300 [W] x 300[D] x 31[H] mmApprox. 2.1 kg (4.7 lb)including batteries*Continua certification is instituted and promoted by PCHA to certify compliance to Continua Design Guidelines. Theguidelines were developed to realize an end-to-end interconnected service for healthcare products.
The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG,Inc. and any use of such marks
by A&D Company,Ltd is under license. Other trademarks and trade names are those of their respective owners.
Android, Google Play™ and the Google Play™ logo are trademarks or registered trademarks of Google LLC.
Apple, the Apple logo and iMessage are trademarks of Apple Inc., registered in the U.S. and other countries. App Store is aservice mark of Apple Inc.
Windows is a trademark or registered trademark of Microsoft Corporation in the U.S. and/or other countries.
Twitter is a registered trademark of Twitter, Inc.
Facebook is a registered trademark of Facebook, Inc.
QR Code is a registered trademark of DENSO WAVE, Inc.
A&D Medical is preparing a range of healthcare devices compatible with A&D Connect app.

Easy-to-use app, a solution for better accuracy
Monitor your blood pressure and track your weight and BMI trends – all with one app from a trusted world-wide leader inconnected health.A&D Connect guides you through the easy device set-up process and provides at-a-glance results,
goals, and trends for you and sharing data with your family and doctors.

A&D Connect

For compatible smartphones and versions, check:
https://link.aandd.jp/Compatibility

Dashboard

List

Graph

Sharing

A&D Company Limited.(JAPAN)

A&D AUSTRALASIA PTY LTD(Australia)

A&D ENGINEERING, INC.(USA)A&D RUS CO., LTD.(Russia)

A&D Instruments Canada Inc.(CANADA)

A&D Technology Trading(Shanghai)Co., Ltd(China)

A&D INSTRUMENTS LIMITED(United Kingdom)

A&D INSTRUMENTS INDIA PRIVATE LIMITED(India)

URL:aandd.jp

URL:andonline.com

URL:autocontrol.com

URL:andprecision.com

URL:andaustralasia.com.au
URL:and-rus.ru

URL:aanddtech.cn
URL:aanddindia.in*BLE series-ADCC-01-IB02-23a00 **