Ambulatory Blood Pressure Monitor TM2440-EX,Instruction Manual, Corrections and Additions
Thank you for purchasing the Ambulatory Blood Pressure Monitor TM2440-EX. The Instruction Manual that
accompanies the product has been corrected and had content added. Please read it in conjunction with thedetails shown below.

Instruction manual Page 22

6.1. Automatic Blood Pressure Measurement (A-BPM)

The following changes have been made.
WrongCorrect

The initial pressurization value is set to 180 mmHg at theThe initial pressurization value for automatic blood pressurefactory.

measurement is AUTO.
In the case of AUTO, the appropriate pressurization value
is set automatically.

Instruction manual Page 37

8.2.3. Initial Pressurization Value

The following changes have been made.
WrongCorrect

The initial pressurization value is set to 180 mmHg at theThe initial pressurization value setting is AUTOfactory.

(appropriate pressurization value set automatically) whenthemark is displayed and 180 mmHg when themark is hidden.

Instruction Manual Page 37

8.3. A-BPM Preset Programs

The following is added after "A-BPM can use only while the automated measurement can be performed. "AdditionNoteThe Start time and Operation time are initially set calculating from thetime at which themarkwasdisplayed.

Whenthenext

automatic measurement is made, this needs to be set again.

1
1WMPD4004497

Instruction Manual Page 40

8.3.1. A-BPM Items and Parameters

The following changes have been made.
WrongCorrect#3: Example for automated measurement.
Start time :Stores a time. (0 to 23 hour)
Operation time : Set to "OFF"
Response :A-BPM starts the blood pressure measurement
at the preset Start time and continues until
A-BPM is suspended.
If the EVENT switch is pressing and holding
until themark is shown, A-BPM starts atthe preset Start time.

#3: If the Start time for automatic measurement is specified#4: Example for automated measurement.
Start time :Set to "OFF"Operation time : Stores time to be continued.
(1 to 27 hours)
Response :A-BPM starts blood pressure measurement
and stops after the Operation time.
If themark is hidden by pressing andholding the EVENT switch during theOperation time, A-BPM stops.
If themark is shown by pressing andholding the EVENT switch again, A-BPM isperformed for the Operation time.

#4: If the Start time for automatic measurement is set to"OFF" and the Operation time is set, long pressing theEVENT switch will implement automatic blood pressure
measurement after initial automatic measurement until
the set Operation time has passed. If the EVENT
switch is long pressed midway through automatic bloodpressure measurement, automatic measurement willstop. If the EVENT switch is long pressed again,
automatic measurement will be performed until the set
Operation time has passed.

Instruction Manual Page 42

and the Operation time is set to "OFF", long pressing
theEVENT

switch will start automatic blood pressure

measurement from the Start time set and measurement
will continue until theIf theEVENT

EVENT

switch is long pressed.

switch is long pressed again, automatic

measurement will continue.
Note
If the Operation time is set, automatic blood pressure
measurement will be performed from when the EVENT
switch is initially long pressed until the Operation time
has passed even if the EVENT switch is long pressed
midway through automatic measurement.

Note
If the Start time is set, long pressing the EVENT
switch midway through automatic blood pressure
measurement will stop automatic measurement and longpressing it again will resume measurement.

8.3.2. A-BPM Program Examples

The following notes were added to "Example 2 Automatic measurement"
Addition
NB. Automatic blood pressure measurement will continue even if turned "OFF" and "ON" again during
automatic measurement.

The following notes were added to "Example 3 Automatic measurement"
Addition
NB. Automatic blood pressure measurement will not be performed once the Operation time has been
exceeded even if turned "OFF" and "ON" again during automatic measurement.

2
1WMPD4004497