Software IRBIS® 3Circuit board

ImageIR® 7300

High-end Thermography Camera – Entry into the VGA Segment
Detector Format
Large detector enables
highest sensivity
640×512100 HzIR-Frame Rate
Analysis of extreme temperaturechanges and gradients in full frame
Measurement Accuracy
Highly accurate andrepeatable measurements
Thermal Resolution
Precise detection of smallest
temperature differences

Those, who are looking for a powerful thermographic camera to solve demanding measurement and testing tasks in thefields of industry and science, that offers an impressive geometrical resolution will find the ImageIR® 7300 a perfect match. Its
cooled focal-plane array photon detector (snapshot) provides(640 × 512) IR pixels and a pitch of 15 μm. In addition, you canchoose between MCT and InSb detectors.
The camera supports fast storing frame rates with frequencies up
to 1,200 Hz. An internal trigger interface guarantees for precise,repeatable triggering of correspondingly fast processes. Two respective inputs and outputs are used to control the camera or togenerate digital control signals for external devices. Dependingon the character of the measurement and testing situation due
to its modular design, most diverse thermographic software andhigh-quality lenses the ImageIR® 7300 is quite easy to adapt tothe on-site conditions.

GigE Vision Compatible
Standard interface for easy integration
into existing process environment
Rotating Filter Wheel and Aperture Wheel
Enables measurement tasks with high object
temperatures and spectral thermography

Focus

Motor Focus
Precise, fast and remotely controlled;
including multiple autofocus functions
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range

MCT: (1.5 … 5.5) µmInSb: (1.5 … 5.7) µm

Pitch

15 µm

Detector

MCT or InSb

Detector format (IR pixels)(640 × 512)

Image acquisition

SnapshotReadout mode

ITR / IWR

Aperture ratio

f/3.0 or f/2.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 300) °C, up to 3,000 °C*

Measurement accuracy± 2 °C or ± 2 %Temperature resolution @ 30 °C

MCT: Better than 0.02 K
InSb: Better than 0.025 K

Frame rate (full / half / sub frame)*

MCT: Up to 75 / 300 / 1,200 Hz
InSb: Up to 100 / 326 / 863 HzWindow mode

Yes* (full frame / sub frame)

Focus

ManualDynamic range

14 bitIntegration time

(1 … 20,000) µs

Rotating filter wheel*

Up to 6 positions

Rotating aperture wheel*

Up to 5 positions

Interfaces

GigE, HDMI*

Trigger

2 IN / 2 OUT, TTL

Analogue signals*, IRIG-B*

2 IN / 2 OUT, noTripod adapter

1/4" and 3/8" photo thread, 2 × M5

Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight

MCT: (241 × 120 × 160) mm*
InSb: (235 × 120 × 160) mm*
3.3 kg (without lens)Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on model

Lenses

Focal length (mm)

FOV (°)

Wide-angle lens12(43.6 × 35.5)

1.3Standard lens25(21.7 × 17.5)

0.6

Telephoto lens50(11.0 × 8.8)

0.3

Telephoto lens100(5.5 × 4.4)

0.15

Telephoto lens

200

(2.7 × 2.2)

0.08

Macro and microscopic lenses

IFOV (mrad)

Minimum object distance (mm)Object size (mm)Pixel size (µm)

Close-up for telephoto lens 50 mm

300(58 × 46)

90

Close-up for telephoto lens 100 mm

500(48 × 38)75

Microscopic lens M = 1.0× (3 versions)

40 / 195 / 300(9.6 × 7.7)1522(3.2 × 2.6)

5

Microscopic lens M = 3.0×
Microscopic lens M = 8.0×× 1.0) and technical progress subject to change without prior notice.
1.9© InfraTec 04 / 2021 – All stated product names and trademarks remain14
in property of their respective owners. Design,(1.2specification