Rotating rotor blade of a wind turbine

ImageIR® 8800

Long Wave Thermography Camera with Shortest Integration Times
Detector Format
Large detector enables
highest sensivity
MicroScan(1,280 × 1,024) IR pixels bygenuine camera hardware

640×512233 HzIR-Frame Rate
Analysis of extreme temperaturechanges and gradients in full frame
Measurement Accuracy
Highly accurate andrepeatable measurements
Shortest Integration Time
Accurate temperature measurements
of fast processes

With its ImageIR® 8800 InfraTec offers another top-level thermographic camera model from the ImageIR® high-end camera
series. It is equipped with a cooled focal-plane array photon
detector that provides a format of (640 × 512) IR pixels andoperates in snapshot mode. This camera combines an outstanding thermal resolution – better than 0.025 K – with very highsub-frame rates of up to 14,593 Hz and extremely short integration times of only a few microseconds. Thereby it qualifies
for airborne biological and geological surveys, non-destructive
testing and the analysis of fast thermal processes, which are related to large temperature measuring ranges. Its modular structure, which consists of optical, detector and interface modules,
makes it easily adaptable to the respective application.
An integrated trigger interface guarantees a repeatable highprecision triggering of quick procedures. Multiple configurable
digital in- and outputs serve as control ports for the camera
or as a generator of control signals for external devices. The
optical channel consists of exchangeable infrared lens systems
as well as of application-specific apertures, filters and opticalelements.

10 GigE Interface
High-speed, long-distance interference
proof data transmission
Spectral RangeMeasurement in the rangeof (7.7 … 10.2) µm
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range(7.7 … 10.2) µm

Pitch

15 µm

Detector

MCTDetector format (IR pixels)(640 × 512)

Image format with opto-mechanical MicroScan (IR pixels)*(1,280 × 1,024)

Image acquisition

SnapshotReadout mode

ITR

Aperture ratio

f/2.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 1,200) °C, up to 3,000 °C*

Measurement accuracy± 1 °C or ± 1 %Temperature resolution @ 30 °C

Better than 0.025 K

Frame rate (full / half / quarter / sub frame)*

Up to 233 / 874 / 2,892 / 14,593 Hz

Window mode

Yes

Focus

Manually, motorised or automatically*

Dynamic range

Up to 16 bit*

Integration time

(10 … 20,000) μs

Rotating aperture wheel and filter wheel*

Up to 5 positions

Interfaces

GigE, 10 GigE*, 2 × CAMLink*, HDMI*

Trigger

4 IN  / 2 OUT, TTL

Analogue signals*, IRIG-B*

2 IN  / 2 OUT, yes

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight(244 × 120 × 160) mm*; 4.0 kg (without lens)

Further functions

Multi Integration Time*

Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on model

Lenses

Focal length (mm)

FOV (°)

Wide-angle lens13(40.5 × 32.9)

1.2Standard lens25(21.7 × 17.5)

0.6

Telephoto lens

50

(11.0 × 8.8)

0.3

Telephoto lens

100

(5.5 × 4.4)

0.15

Telephoto lens200(2.7 × 2.2)

0.08

Observation of a person entering an area without authorisation

IFOV (mrad)

Thermal image of a drinking bat over a reflecting water surface© InfraTec 04 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.