NEW

Impact of a water drop on a soldering iron, recorded
in high-speed mode

ImageIR® 9400 hp

High-speed Infrared Camera in HD Format

Detector Format
Efficient measurement of smallest
structures on large-scale objects
MicroScan(2.560 × 2.048) IR pixels bygenuine camera hardware
640×512622 Hz

IR-Frame Rate
Analysis of extreme temperaturechanges and gradients in full frame
Measurement Accuracy
Highly accurate andrepeatable measurements
Thermal Resolution
Precise detection of smallest
temperature differences
10 GigE Interface
High-speed, long-distance interference
proof data transmission
High-speed Mode
Increase frame rates and thermal resolution
at the same time using binning technology

The strengths of the ImageIR® 9400 hp infrared camera from
InfraTec are based on its versatility. It is the most universal camera of the series and offers both a high frame rate and an
extremely high geometric resolution of up to (1,280 × 1,024) IR
pixels. The high thermal resolution of the camera allows the detection of the smallest temperature differences as low as 20 mK(0.02°C) at 30°C. It measures temperature differences whereother cameras only record noise. Based on binning technology,
a high-speed mode can be selected in addition to the standard
mode. It allows the frame rate to be tripled while maintaining
the same field of view. At the same time, the thermal resolution increases by a factor of 2. Due to the opto-mechanicalMicroScan, the geometric resolution can be increased to image
formats with up to (2,560 × 2,048) IR pixels. This results in anenormous gain in measurement data that can be used for further analysis. In addition, the image quality is significantly improved, allowing even the finest structures to be resolved withenormous sharpness of detail.
The modular design enables individual configuration and consistent alignment to highly demanding applications in scienceand industry. Depending on the measurement task, HighSense
allows users to select individual integration times while maintaining calibration. Interchangeable precision lenses, in combination with the motorised focus function, enable fast, precise
and remote motorised focusing as well as a reliable autofocus
function.

www.InfraTec-infrared.com

www.InfraTec.deTechnical Specifications
Spectral range(1.5 … 5.5) µm

Pitch10 µm

DetectorInSb

Detector format (IR pixels)(1,280 × 1,024)

Detector format with opto mechanical MicroScan (IR pixels)(2,560 × 2,048)

Image acquisition

SnapshotReadout mode

ITR / IWR

Aperture ratio

f/2.2 or f/3.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 1,500) °C, up to 3,000 °C*

Measurement accuracy± 1 °C or ± 1 %Temperature resolution @ 30 °C

Better than 0.03 K / 0.02 K with High-speed Mode

Frame rate (full / half / quarter / sub frame)*

Up to 180 / 342 / 622 / 2,601 HzHigh-speed Mode: up to 622 / 1,053 / 1,615 / 3,343 Hz

Window mode

YesFocus

Manual, motorised or automatic*

Dynamic range

Up to 16 bit*

Integration time(1 … 20,000) μs

Rotating filter wheel*

Up to 6 positions

Rotating aperture wheel*

Up to 5 positions

Interfaces

10 GigE, HDMI*, GigE*, CamLinkTrigger

4 IN  / 2 OUT, TTL

Analogue signals*, IRIG-B*

3 IN  / 2 OUT, yes

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5

Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight(241 × 123 × 160) mm*; 4.3 kg (without lens)

Further functions

High-speed Mode*, Multi Integration Time*, HighSense*

Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on modelLenses

Focal length (mm)

FOV (°)

Standard lens25(29 × 23)

IFOV (mrad)
0.4

Telephoto lens50(15 × 12)

0.2

Telephoto lens100(7.3 × 5.9)

0.1

Telephoto lens200(3.7 × 2.9)

0.05

Minimum object distance (mm)Object size (mm)Pixel size (µm)

Macro andmicroscopic lenses
Close-up for telephoto lens 50 mm

300(77 × 61)

60

Close-up for telephoto lens 100 mm

500(64 × 51)50

Microscopic lens M=1.0×40(13 × 10)10Microscopic lens M=2.5×14(5.1 × 4.1)

4Microscopic lens M=8.0×14(1.6 × 1.3)

1.3

Fields of application:
 Monitoring of fast processes Material testing Microthermography
 Research and development

Detecting the smallest temperature differences withthe highest level of detail using MicroScan© InfraTec 08 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.