Tester for Low Beam
Color Image
Processing Method

Manual Headlight Tester
Real color Image Processing
HLI−215HEADLIGHT TESTERMODELEvolving Image Processing

Latest Technology

RGB Color Method

High performance headlight tester equippedwith a color CCD camera to determine the color
shade that supports various lights
and light sources
Using Latest Technology RGB Color Method

Color CCD Camera

Can perform analysis close to that of human eyes by taking
advantage of the characteristics of a color camera to determine
the color shade of the light from the balance of the RGB values.
It enhances the ability to respond to increasingly diverse new lightsources and improves the measurement speed and accuracy.
The color image processing headlight tester

detects the RGB of thelight of the headlightwith a color CCD camera

determines the color andconverts the acquired
RGB values to brightness
intensity levels

displays the accurate
luminosity/irradiation
direction

Chromaticity Display Function to Allow You toEasily Determine the Color Shade of Light
Equipped with large
17-inch monitor
Support for fog lightmeasurement height 25 cm

Low beam
New light
sources suchas HID andLED
Chromaticity display screen(CIE chromaticity diagram)

JIS chromaticity screen (JIS Z8701)

Lightdistribution
pattern of aZ-beam, etc.
Fog light

Adjust mode Screen mode

Light Receiver Moves Upand Down Electrically
The light receiver and the monitor move up
and down electrically. Hold down thebutton to increase the speed in the upward
movement. The speed can be varied.

Traverse handle

Large Easy-to-View Alignment Screen
Light alignment screen arranged in front of the lightreceiver allows you to align the light more accurately.

Useful Functions■ External Dimensions■ Screen Mode
Adjustment to an arbitrary
position can be performed byplacing the virtual cutline(adjustment target) and aligning
the light image with it.

(719.2)2413451728

1838 (When MAX1200)■ Adjust Mode
The target can be moved to anyposition for a special light for
which the elbow point is hard toacquire. When the position isdetermined, the optical axis canbe adjusted as the target follows
during light adjustment.

70280

500 MAX.1200
60MIN.250

420

F.L

F.L

Easy-to-read Measurement Results ScreenStandard rail

420626802870.5■ Option

Measurement results can be checked all at once on the measurementresults screen. Measurement data can be stored by pressing anoperation button so work efficiency is substantially improved.

Forward and Backward Moving Device(Specially manufactured device)■ The tester automatically detects the distance to the headlight of the vehicle and moves forward or backward to
the measurement distance (1 m).
■ Since the measurement can be taken without moving the vehicle, the work efficiency can be improved.
■ The vehicle inspection line space also can be saved by using this device in conjunction with the 4WD-BS tester.
* For details, please contact one of our sales offices.

Vehicle Alignment Laser PointerLow-beam light/high-beam lightmeasurement results screen

CD:01210152

Fog light measurement
results screen■ Specifications

Inspection Flow at Service Shop

Model
MLIT registered model

HLI−215
HLI−2015

As a result of a revision of the safety standards and review procedure rules, the headlight testof vehicles produced on and after September 1, 1998, will be changed to a low-beam headlighttest in September 1, 2015.

Model test number

JASEA−H−39

Was the vehicle produced on or after September 1, 1998?

Measurement method

Manual/light condensation

Measurement distance(cm)

25 to 120

High-beam light(hcd)

0 to 1,200

Luminosity Low-beam light(hcd)

0 to 1,200

Fog light(hcd)

0 to 1,200

Measurement range

Light mounting height measurement range

High-beam light/low-beam light
HighLeftRight(cm/10m)

35

Yes

Yes

No

Do the test results of the low-beam headlight conform to the standards?

35

Yes

High 20.0 to 0 to low 35.0

Luminosity/irradiation direction

LCD digital

Light distribution/alignment

Image/screen method

Power supply(V/A)

AC100/5 50/60Hz

Tester dimensions(mm)

W870.5×D626×H1,728

Tester weight(kg)Approx. 80

Standard accessories

NoHigh-beam headlight tester: Can thelow-beam headlight be tested using a screen?
(Is it possible to measure by projecting thelight directly onto a wall, etc.?)

35

Low

No

Is the headlight tester a low-beam headlight tester?

20

Fog light
Display type(Except for motorcycles, motorcycles with sidecar, etc.)

Yes

1(m)

No
Adjustment by service person
Difficult to judge
Do the test results of the high-beam headlight conform to the standards?

Yes

No
Adjustment by service person

Auxiliary light cover x1, light receiver cover x1, convex rule x1

Standard rail dimensions(mm)

W420×L4,500

Applicable rail width(mm)360, 480, 550, 600■ Before using this product, carefully read the precautions indicated byDANGER,

Pass* For details, visit the MLIT website.

WARNING, andCAUTION in the manual supplied with this product to ensure correct use.

4-16-25 Shibaura, Minato-ku, Tokyo 108-0023
Phone: +81 3-5441-3412 Fax: +81 3-5441-8848
ANZEN website: http://www.anzen.co.jp

CAT.1.0'1512_010418_2618
HLI–215

CD:01212067* The appearance and specifications are subject to change without notice.