Heat development during the ABS brake process

Software IRBIS® 3 rotate for rotation test bench

ImageIR® 5300

High-speed Thermography Camera with Large Pitch

Detector Format
Large detector enables
highest sensivity

105 kHz

IR-Frame Rate
Analysis of extreme temperaturechanges and gradients
Measurement Accuracy
Highly accurate andrepeatable measurements
Thermal Resolution
Precise detection of smallest
temperature differences
Shortest Integration Time
Accurate temperature measurements
of fast processes
Pitch Dimension
Precise measurement of low temperaturesand very fast integration times

The ImageIR® 5300 has been designed specifically for capturing and recording extremely fast running thermal processes.
The MWIR focal-plane array photon detector in the format of(320 × 256) IR pixels allows users to capture thermal images
in full frame at frequencies up to 481 Hz. When using the subframe mode, the value even increases to 105,000 Hz.
The ImageIR® 5300 demonstrates the strength of its design as
an integral part of the automated IR rotation test bench solution
Thermal Rotate Check (TRC) from InfraTec. This allows rapidly
rotating components, such as tyres, brakes and clutches, to beanalysed precisely. The results provide information on how well
the test objects withstand continuous operation, which signs ofwear are present and how serious they are.
The potential of the camera goes far beyond such applicationsin automotive and rail technology. Thanks to its extensive single pixels (detector pitch 30 µm) the ImageIR® 5300 achieves anoutstanding thermal resolution better than 0.015 K. Modularly designed with an optics, detector and interface module andequipped with an integrated trigger interface, the camera provesitself to be a versatile measuring and testing instrument for applications in industry and science.

GigE Vision Compatible
Standard interface for easy integration
into existing process environment
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range(3.7 … 4.8) µm

Pitch

30 µm

Detector

MCT

Detector format (IR pixels)(320 × 256)

Image acquisition

SnapshotReadout mode

ITR

Aperture ratio

f/2.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 1,200) °C, up to 3,000 °C*

Measurement accuracy± 1 °C or ± 1%

Temperature resolution @ 30 °CBetter than 0.015 K

Frame rate (full / half / quarter / sub frame)*

Up to 481 / 1,906 / 7,229 / 105,000 Hz

Window mode

Yes

Focus

Manual, motorised or automatically*

Dynamic range

Up to 16 bit*

Integration time

(1 … 20,000) µs

Rotating filter wheel*

Up to 5 positions

Rotating aperture wheel*

Up to 5 positions

Interfaces

GigE, HDMI*

Trigger

4 IN / 2 OUT, TTL

Analogue signals*, IRIG-B*

2 IN / 2 OUT, yes

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight(241 × 120 × 160) mm*; 3.3 kg (without lens)

Further functions

Multi Integration Time*

Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 rotate, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3control*, IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*,IRBIS® 3 vision** Depending on modelLenses

Focal length (mm)

FOV (°)IFOV (mrad)

Wide-angle lens12(43.6 × 35.5)

2.5Standard lens25(21.7 × 17.5)

1.2

Telephoto lens

50

(11.0 × 8.8)

0.6

Telephoto lens

100

(5.5 × 4.4)

0.3

Telephoto lens

200

(2.7 × 2.2)

0.15

Macro and microscopic lenses

Object distance (mm)Object size (mm)Pixel size (µm)

Close-up for telephoto lens 50 mm

300(58 × 46)

180

Close-up for telephoto lens 100 mm

500(48 × 38)150

Microscopic lens M=1.0× (2 versions)

195 /300(9.6 × 7.7)3022

(3.2 × 2.6)

10Microscopic lens M=3.0×© InfraTec 04 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.