FG-D-CWP / FG-D-ACWP(This document is included in the packing contents for the balance.)

English (EN)
Quick Start Guide
For Water proof platform scale ........................................................ 3Français (FR)Guide de démarrage rapide
Pour balance plate-forme étanche .................................................. 9
Deutsch (DE)
Schnellstartanleitung
Für wasserfeste Plattformwaage ................................................... 15
Español (ES)
Guía de inicio rápido
Para báscula de plataforma a prueba de agua ............................. 21
Italiano (IT)Guida introduttiva
Per bilancia a piattaforma impermeabile ....................................... 27
中文 (CT)
快速入門指南防水台秤用 ..................................................................................... 33

1WMPD4004769C

Warning Definition
The warning definition described in this manual has the following meaning:

DANGERAN imminently hazardous situation which, if not avoided, will
result in death or serious injury.

© 2022 A&D Company, Limited. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated into
any language in any form by any means without the written permission of A&D
Company,Limited.

The contents of this publication and the specifications of the instrument covered by thispublication are subject to change for improvement without notice.

EN
Quick start guide (EN)

1. Introduction
This quick start guide describes the basic functions and operations of this product.
The FG-D-CWP / FG-D-ACWP weighing indicator may be used on a waterproof platform scale with IP67
performance. It has similar specifications and functions as an indicator of the FG-CWP series.
Detailed information about FG-D-CWP / FG-D-ACWP is provided in the following instruction manuals.
These manuals can be downloaded from the A&D website (https://www.aandd.jp).
Scan this QR code for this manual.

[FG-D-CWP / FG-D-ACWP Instruction Manual]This is an instruction manual utilized when connecting and installing the FG-D-CWP / FG-D-ACWP to aweighing platform (load cell).
[FG-CWP / FG-ACWP Series Instruction Manual]This is an instruction manual for understanding and fully utilizing the waterproof digital platform scale FGCWP series.

2. Items Included
The following items are packed with the product.
FG-D-CWP (USB power model)
FG-D-CWP

FG-D-ACWP (AC power model)
FG-D-ACWP

Model label(1083039230-1)

Attach here

Write the scale’s
ratings, etc.

Attach here

Model label(1083039230-3)

Write the scale’s
ratings, etc.

Indicator

Indicator

USB cable(Approx. 3 m)

AC cable(Approx. 3 m)

AC adapter(Image diagram)
FG-D-CWP / FG-D-ACWP common accessories
Quick start guide

Rubber packing(To be used when the diameterof load cell cable is from φ3.5 to
φ7.0)

3

Rubber feet for display stand (4 pcs.)

EN

3. Cautions
3.1. Precautions for Installing the Indicator
DANGER
Ground the scale, so that the user will not be subjected to an electric shock.
To prevent electric shock, do not handle the USB cable / AC cable with wet hands.
The USB plug is not water-resistant. Use a USB outlet located at a place where the plug will not get wet.
(For FG-D-CWP (USB power model))The AC plug is not water-resistant. Use an electrical outlet located at a place where the plug will
not get wet. (For FG-D-ACWP (AC power model))Do not install the scale where flammable or corrosive gas is present.
Do not put excessive force on cables.
The scale is heavy. Use caution, as lifting may cause it to fall over.
Use shielded load cell cables.
Please read "Precautions for Installing the Scale" in [Instruction Manual: FG-CWP / FG-ACWP Series]
on the A&D website (https://www.aandd.jp) for other precautions, cautions as an assembled scale, storage and cleaning.
Please read "Installation and Precautions" in [Instruction Manual: FG-D-CWP / FG-D-ACWP] on the A&D website (https://www.aandd.jp) regarding connection of load cells, installation of indicators, functio
n setting and sensitivity adjustment, etc.

3.2. Load Cell Connection
FG-D-CWP / FG-D-ACWP has two connection configurations: the 6-wire configuration and the 4-wireconfiguration.
We recommend connecting using the 6-wire configuration for optimum accuracy and stability.
The applicable diameter of load cell cable is from φ3.5 to φ9.0 mm.
The length of the load cell cable should be less than 30m. For 4-wire configuration, we recommend using
a length of 5m or less.
Terminal No.
1
2
3
4
5
6
7

SIGSIG+
EXCSENSEN+EXC+
SHIELD

Function
Load cell input (-)
Load cell input (+)
Load cell excitation voltage (-)
Sensing input (-)
Sensing input (+)
Load cell excitation voltage (+)
Shield4

EN
Load cell

Cable

Cable

Load cell input (-)
Load cell input (+)
Load cell excitation voltage (-)
Sensing input (-)
Sensing input (+)
Load cell excitation voltage (+)
ShieldShield(A) 6-wire configuration connection(B) 4-wire configuration

to load cell (recommended)

connection to load cell

4. Part Names<FG-D-CWP (USB power model)>Display Unit (SUS304)
Display unit knob
Serial Number

Display Stand /Wall-Mount Bracket

Load Cell Panel
Load Cell Connector
USB Cable (Approx. 3 m)
USB Plug<FG-D-ACWP (AC power model)>Display Unit (SUS304)

Display unit knob
Serial Number

Load Cell Panel

Display Stand /Wall-Mount Bracket

Load Cell Connector
AC Cable (Approx. 3 m)
Earth
5Earth Terminal

EN

4.1. Display and Symbols
Display Unit
Weighing
Condition
Comparator LED
Conditions for
Each Function

Weighing Units

Display/symbol

Description
This is lit when the weighing value is stable, indicating that the scale is in the propercondition for reading weighing values.

STABLENETThis is lit when weight of the container (tare) is subtracted.

ZERO

This is lit when the scale is at the zero point (reference point for weighing).

Weighing Units

“

”, “

”, “

”, “

”,“

” and “” is lit.

This is lit when the display is held.
This is a function to detect impact to the mass sensor section and to display the impact
level.
Turns on when the connection with the wireless communication receiver is established.
While the comparator function is being used, the weighing value is compared using thepreset threshold values and the indicator displays the result.

Alternating current.
Earth (ground) Terminal.
For details on the method of operation, various functions, function table, etc. refer to the [Instruction
Manual: FG-CWP / FG-ACWP Series] on our website (https://www.aandd.jp).
However, please note that functions related to USB and mobile battery cannot be used with FG-DACWP (AC power model).

4.2. Operation Keys
Operation key

Description
ON/OFF Key
Shows or hides the display.
SAMPLE Key
Selecting “pcs” as the unit activates individual weight setting mode.
Pressing and holding (for about 2 seconds) activates the comparator threshold setting
mode.

6

EN
Operation key

Description
PRINT Key
Outputs the weighing value as data.
Increases the value of the digit flashing when setting.
MODE Key
Switches the weighing unit.
Shifts to the right the digit flashing when making settings.
ZERO Key
Based on the zero point weighed at the time of turning on the display, by pressing theZERO key when the weight value is stable within ±2 % of the weighing capacity, the zero
point indicator turns on as soon as the display becomes zero.
TARE KeyThe weight of the item on the weighing pan is subtracted as the weight of a container (tareweight).

5. Maintenance･Refer to "3. Cautions" regarding use.
･Refer to "5.3. Error Code Table" and corresponding mode for displayed error code.
･Refer to the [FG-CWP / FG-ACWP Series Instruction Manual] on the A&D website (https://www.aandd.jp) for
sensitivity adjustment.
･Periodically check the accuracy of weighing. Adjust the scale, if it is moved to another location or theenvironment has changed.

5.1. Repair
Do not disassemble / assemble the scale without an authorized service engineer. Doing so may cause anelectric shock or damage to the scale, etc. In this case, repair is not covered under warranty. Contact yourlocal A&D dealer if your scale needs service or repair.

5.2. Error display
In this situation

Confirm these items

Nothing displayed.
Scale does not turn on.

･Is the main power the correct voltage?
･Is the main power cord properly connected?

The scale does not display
zero at first.

･Check around the weighing pan.
･Is there anything on the weighing pan?
･Refer to the [FG-D-CWP / FG-D-ACWP Instruction Manual] on the A&D
website (https://www.aandd.jp) and perform zero point sensitivity
adjustment.

is displayed.

･Weighing error that meaning "Over loaded".

is displayed.

･Indicates that the sample weight is too light to set the unit weight in the
counting mode.

is displayed.

･Sensitivity adjustment error that meaning "Too light".

is displayed.

･Displayed when the main power drops.

7

EN
In this situationisdisplayed and does notproceed.

Fixed display.

Confirm these items･The weighed value is unstable due to drift, vibration or other factors.
A breeze or vibration may be affecting measurement.
Check around the weighing pan.
･Check the connection of the load cell cable.
･No zero display when the display is turned on.
Remove anything that is on the pan.
･Refer to the [FG-D-CWP / FG-D-ACWP Instruction Manual] on the A&D
website (https://www.aandd.jp) and perform zero point sensitivity
adjustment.
･Did you use the "hold function" ?
･Refer to the [FG-CWP / FG-ACWP Series Instruction Manual] on the A&D
website (https://www.aandd.jp) and change the function.
･Turn off scale and turn it on again.

5.3. Error Code Table
When any of the following errors are displayed, try turning the display off and on again.

Display

Content

is displayed

The temperature sensor has failed.

is displayed

The memory (circuit) has failed.

is displayed

The internal circuitry has failed.

is displayed

The mass sensor has failed.

6. Specifications
Specifications are subject to change without notice.
Input sensitivity
0.1 μV/d min. (d: minimum division)
Input signal range-16 mV to 16 mV
Load cell excitation voltage
5 V ± 10 %Load cell drive capacity
Up to 4 x 350 ohm loadcells
Temperature coefficient Zero
±0.02 μV/°C typ. ±0.1 μV/°C max.
Temperature coefficient Span ±3 ppm/°C typ. ±15 ppm/°C max.
linearity±0.01 % of full scale
Maximum display resolution
100,000
Display
7 segment LCD with backlight (Character height 30 mm)
Display update
Approx. 10 times/second
Operating Conditions-10 °C to 40 °C, 85 % R.H. or less (non-condensing)
Installation environment
Indoor use only
Altitude
Altitude up to 2,000 m
Power Supply<FG-D-CWP (USB power model)>Power supplied from AC adapter, mobile battery, or USB Type-A port
USB Cable Length: Approx. 3 m(USB cable is also used for communication.)<FG-D-ACWP (AC power model)>AC mains (100 V to 240 V, 50/60 Hz, 0.1 A Max)
AC Cable Length: Approx. 3 m
Overvoltage category *II
Pollution degree *2
Dimensions
Width 240 mm × Depth 76 mm × Height 155 mm
Weight (approximately)
1.6 kg (Includes stand)Dust and water protection
Complies with IP67

* Applicable to FG-D-ACWP (AC power model) only

8

FR
Guide de démarrage rapide (FR)

1. Introduction
Ce guide de démarrage rapide décrit les fonctions et les opérations de base de ce produit.
L'indicateur de pesage FG-D-CWP / FG-D-ACWP peut être utilisé sur une balance plate-forme étanche
conforme à l'indice de performance IP67. Ses spécifications et fonctions sont similaires à celles del'indicateur de la série FG-CWP.
Des informations détaillées sur FG-D-CWP / FG-D-ACWP sont fournies dans le manuel d'instructions suivant.
Ces manuels peuvent être téléchargés depuis le site web d'A&D (https://www.aandd.jp).
Scannez ce code QR pour ce manuel.

[Manuel d'instructions de FG-D-CWP / FG-D-ACWP]
Ce manuel d'instructions doit être utilisé lors du raccordement de FG-D-CWP / FG-D-ACWP à une plateforme de pesage (cellule de charge) et de son installation sur cette dernière.
[Manuel d'instructions de la série FG-CWP / FG-ACWP]
Ce manuel d'instructions doit permettre de comprendre et d'exploiter pleinement la balance plate-forme
numérique étanche de la série FG-CWP.

2. Articles inclus
Le produit est emballé avec les articles suivants.
FG-D-CWP (Modèle à alimentation USB)
FG-D-CWP

Fixer ici

FG-D-ACWP (Modèle à alimentation secteur)
FG-D-ACWP

Étiquette de modèle(1083039230-1)

Écrire les indices dela balance, etc.

Étiquette de modèle(1083039230-3)

Fixer ici

Indicateur

Écrire les indices dela balance, etc.

Indicateur

Câble secteur(environ 3 m)

Câble USB
(environ 3 m)

Adaptateur CA(Schéma illustratif)
Accessoires communs FG-D-CWP / FG-D-ACWP
Guide de démarrage rapide
Emballage en caoutchouc(diamètre des câbles de la cellulede charge / utilisé en cas
d'utilisation de Φ3,5 à Φ7,0)

9

Pieds en caoutchouc pour lesupport d'affichage (4 pièces)FR

3. Avertissements
3.1. Précautions d'installation de l'indicateur
DANGER
Mettre la balance à la terre afin de protéger l'utilisateur des chocs électriques.
Afin d'éviter les chocs électriques, ne pas manipuler le câble USB/secteur avec les mains
mouillées.
La fiche USB n'est pas résistante à l'eau. Utiliser une prise USB située à un endroit où la fiche ne sera pas
mouillée (pour FG-D-CWP (modèle à alimentation USB)).
La fiche secteur n'est pas résistante à l'eau. Utiliser une prise électrique située à un endroit où lafiche ne sera pas mouillée. (pour FG-D-ACWP (modèle à alimentation secteur))Ne pas installer la balance dans un environnement où des gaz inflammables ou corrosifs sontprésents.
Ne pas soumettre les câbles à une force excessive.
La balance est lourde. Faire preuve de prudence lorsque la balance est soulevée, car elle risquede basculer.
Utiliser des câbles de cellule de charge blindés.
Lire « Précautions d'installation de la balance » dans [Manuel d'instructions : série FG-CWP / FG-ACWP] surle site web d'A&D (https://www.aandd.jp) pour plus de précautions et d'avertissements sur le montage, lestockage et le nettoyage de la balance.
Lire « Installation et précautions » dans [Manuel d'instructions : FG-D-CWP / FG-D-ACWP] (sur le site web
d'A&D (https://www.aandd.jp) pour le raccordement de cellules de charge, l'installation d'indicateurs, leréglage des fonctions et de la sensibilité, etc.

3.2. Raccordement de cellules de charge
FG-D-CWP / FG-D-ACWP dispose de deux configurations de raccordement : la configuration à 6 fils et laconfiguration à 4 fils.
Nous recommandons de raccorder en configuration à 6 fils pour une précision et une stabilité optimales.
Le diamètre applicable du câble de la cellule de charge est compris entre Φ3,5 et Φ9,0 mm.
La longueur du câble de la cellule de charge doit être inférieure à 30 m. Pour une configuration à 4 fils,
nous recommandons d'utiliser une longueur de 5 m ou moins.
No de borne
1
2
3
4
5
6
7

SIGSIG+
EXCSENSEN+EXC+
SHIELD

Fonction
Entrée de la cellule de charge (-)
Entrée de la cellule de charge (+)
Tension d'excitation de la cellule de charge (-)
Entrée de détection (-)
Entrée de détection (+)
Tension d'excitation de la cellule de charge (+)
Blindage

10

FR
Cellule de charge

Câble

Câble

Entrée de la cellule de charge (-)
Entrée de la cellule de charge (+)
Entrée de détection (-)
Entrée de détection (+)
Blindage

Blindage(A) Raccordement en configuration à 6 fils
à une cellule de charge (recommandé)(B) Raccordement en configuration
à 4 fils à une cellule de charge

4. Noms des pièces<FG-D-CWP (Modèle à alimentation USB)>
Unité d'affichage (SUS304)

Molette de l'unitéd'affichage
Numéro de série

Panneau de lacellule de charge

Support d'affichage /Support muralConnecteur de lacellule de charge
Câble USB (environ 3 m)
Prise USB

<FG-D-ACWP (Modèle à alimentation secteur)>
Molette de l'unitéd'affichage

Display Unit (SUS304)

Numéro de série

Support d'affichage /Support muralPanneau de lacellule de charge
Connecteur de lacellule de charge
Câble secteur (environ 3 m)
Terre
11

Borne de terre

FR

4.1. Affichage et symboles
Unité d'affichage

Conditionde pesage
LED de comparateur

Conditions dechaque fonction

Unités de pesage

Affichage/symbole
STABLENET
ZERO
Unités de pesage

Description
Ceci s'allume quand la valeur de pesage est stable afin d'indiquer que la balancedispose des bonnes conditions pour lire les valeurs de pesage.
Ceci s'allume quand le poids du contenant (tare) est soustrait.
Ceci s'allume quand la balance se trouve au point zéro (point de référence pour lepesage).
«

», «

», «

», «

», «

» ou « » s'allume.

Ceci s'allume quand l'affichage est tenu.
Cette fonction sert à détecter l'impact sur la section du capteur de masse et à
afficher le niveau d'impact.
Ceci s'allume quand la connexion est établie avec le récepteur de communicationsans fil.
Lorsque la fonction de comparateur est utilisée, la valeur de pesage est comparéeà l'aide des valeurs de seuil prédéfinies et l'indicateur affiche le résultat.
Courant alternatif
Borne de terre
Pour plus de détails sur la méthode d'utilisation, les différentes fonctions, le tableau des fonctions,etc., se référer au [Manuel d'instructions : série FG-CWP / FG-ACWP] sur notre site Web(https://www.aandd.jp).
Toutefois, veuillez noter que les fonctions liées à l'USB et à la batterie mobile ne peuvent pas êtreutilisées avec le FG-D-ACWP (modèle à alimentation secteur).

4.2. Touches de commandeTouche decommande

Description
Touche ON/OFF
Affiche ou masque l'affichage.
Touche SAMPLE
Lors de la sélection de « pcs » comme unité, le mode de réglage individuel du poids
est activé.
Une pression longue (d'environ 2 secondes) active le mode de réglage du seuil ducomparateur.
12

FR
Touche decommande

Description
Touche PRINT
Fournit la valeur de pesage sous forme de données.
Augmente la valeur du chiffre qui clignote lors du réglage.
Touche MODE
Change l'unité de pesage.
Déplace vers la droite le chiffre qui clignote lors du réglage.
Touche ZERO
D'après le point zéro pesé lors du démarrage de l'affichage, lors d'une pression sur
la touche ZERO quand la valeur du poids est stable à ±2 % de la capacité depesage, l'indicateur du point zéro s'allume dès que l'affichage passe à zéro.
Touche TARE
Le poids de l'article posé sur le plateau de pesage est soustrait en tant que poids decontenant (tare).

5. Maintenance･Se référer à « 3. Avertissements » pour l'utilisation.
･Se référer à « 5.3. Tableau des codes d'erreur » et au mode correspondant pour le code d'erreur affiché.
･Se référer à [Manuel d'instructions de la série FG-CWP / FG-ACWP] sur le site web d'A&D (https://www.aandd.jp)
pour le réglage de la sensibilité.
･Vérifier régulièrement la précision du pesage. Régler la balance si elle a été déplacée dans un nouvel
endroit ou si l'environnement a changé.

5.1. Réparation
Ne démonter/monter la balance qu'en présence d'un technicien d'entretien agréé. Sinon, l'utilisateur risquede subir un choc électrique ou d'endommager la balance, etc. Dans ce cas, la garantie ne couvre pas laréparation. Contacter le distributeur A&D local si la balance a besoin d'un entretien ou d'une réparation.

5.2. Affichage d'erreurs
Dans cette situation
Rien ne s'affiche.
La balance ne démarre pas.
La balance n'affiche pas lezéro en premier.
s'affiche.

Vérifier ces éléments･L'alimentation principale est-elle à la bonne tension ?
･Le cordon de l'alimentation principale est-il correctement raccordé ?
･Vérifier les abords du plateau de pesage.
･Y a-t-il quelque chose sur le plateau de pesage ?
･Se référer à [Manuel d'instructions de FG-D-CWP / FG-D-ACWP] sur lesite web d'A&D (https://www.aandd.jp) et régler la sensibilité du point
zéro.
･Erreur de pesage signifiant « Surcharge ».

s'affiche.

･Indique que le poids d'échantillon est trop léger pour régler le poids del'unité en mode de comptage.

s'affiche.

･Erreur de réglage de la sensibilité signifiant « Trop léger ».

s'affiche.

･S'affiche lorsque l'alimentation principale chute.

s'affiche
et l'affichage ne changepas.

･La valeur pesée est instable en raison d'un décalage, de vibrations ou
d'autres facteurs.
Une brise ou des vibrations perturbent peut-être la mesure.
Vérifier les abords du plateau de pesage.
･Vérifier le raccordement du câble de la cellule de charge.
13

FR
Dans cette situation

Affichage fixe.

Vérifier ces éléments･Le zéro ne s'affiche pas lorsque l'affichage est démarré.
Retirer tout article éventuellement posé sur le plateau.
･Se référer à [Manuel d'instructions de FG-D-CWP / FG-D-ACWP] sur lesite web d'A&D (https://www.aandd.jp) et régler la sensibilité du pointzéro.
･La fonction « Hold » a-t-elle été utilisée ?
Se référer à [Manuel d'instructions de la série FG-CWP / FG-ACWP] sur lesite web d'A&D (https://www.aandd.jp) et modifier la fonction.
･Arrêter la balance, puis la redémarrer.

5.3. Tableau des codes d'erreur
Lorsque l'une des erreurs suivantes s'affiche, essayer d'arrêter l'affichage et de le redémarrer.

Affichage

Contenus'affiche

Le capteur de température a dysfonctionné.

s'affiche

La mémoire (circuit) a dysfonctionné.

s'affiche

Les circuits internes ont dysfonctionné.

s'affiche

Le capteur de masse a dysfonctionné.

6. Spécifications
Les spécifications sont sujettes à modifications sans préavis.
Sensibilité d'entrée
0,1 μV/d min. (d : division minimale)Plage du signal d'entrée-16 mV à 16 mV
Tension d'excitation de cellule 5 V ± 10 %de charge
Capacité d'excitation deJusqu'à 4 cellules de charge de 350 ohms
cellule de charge
Coefficient de température au ±0,02 μV/°C typ. ±0,1 μV/°C max.
point zéro
Coefficient de température sur ±3 ppm/°C typ. ±15 ppm/°C max.
l'intervalle
Linéarité±0,01 % de la pleine échelle
Résolution d'affichage
100,000
maximum
Affichage
LCD 7 segments avec rétroéclairage (hauteur des caractères : 30 mm)
Actualisation de l'affichage
Environ 10 fois/secondeConditions de fonctionnement -10 °C à 40 °C, 85 % HR ou moins (sans condensation)
Environnement d'installation
Utilisation en intérieur uniquement
Altitude
Altitude jusqu'à 2 000 m
Alimentation<FG-D-CWP (modèle à alimentation USB)>
Alimentation fournie par l'adaptateur secteur, la batterie mobile ou le port
USB de type A.
Longueur du câble USB : environ 3 m(Le câble USB est également utilisé pour la communication.)<FG-D-ACWP (modèle à alimentation secteur)>
Courant secteur (100 V à 240 V, 50/60 Hz, 0.1 A Max)
Longueur du câble secteur : environ 3 m
Catégorie de surtension *II
Degré de pollution *2
Dimensions
240 mm de largeur × 76 mm de profondeur × 155 mm de hauteur
Poids (approximatif)
1,6 kg (socle inclus)
Protection contre l’eau et laConforme à la norme IP67
poussière* Applicable uniquement au FG-D-ACWP (modèle à alimentation CA)
14

DE
Schnellstartanleitung (DE)

1. Einführung
In dieser Schnellstartanleitung werden die Grundfunktionen und die Bedienung des Produkts
beschrieben.
Die Wägeanzeige FG-D-CWP / FG-D-ACWP kann auf einer wasserfesten Plattformwaage mit IP67Leistung verwendet werden. Die Spezifikationen und Funktionen ähneln denen der Serie FG-CWP.
In den folgenden Bedienungshandbüchern finden Sie detaillierte Informationen über FG-D-CWP / FG-DACWP. Diese Handbücher können von der A&D-Webseite (https://www.aandd.jp) heruntergeladen werden.
Scannen Sie diesen QR-Code für dieses Handbuch.

[FG-D-CWP / FG-D-ACWP-Bedienungshandbuch]
Dieses Handbuch wird zur Verbindung und Installation von FG-D-CWP an eine Wägeplattform(Wägezelle) verwendet.
[Bedienungshandbuch FG-CWP/ FG- ACWP-Serie]Dieses Handbuch dient dem Verständnis und der vollständigen Verwendung der wasserdichten digitalen
Plattformwaage der FG-Serie.

2. Im Packungsumfang enthalten
Die folgenden Elemente werden mit dem Produkt verpackt.
FG-D-CWP (USB-Strom-Modell)FG-D-CWP

FG-D-ACWP (Netzstrom-Modell)FG-D-ACWP

Modelletikett(1083039230-1)

Schreiben Sie die
Werte der Waage
usw. auf.

Hier anbringen

Hier anbringen

Modelletikett(1083039230-3)

Schreiben Sie die
Werte der Waage
usw. auf.

Anzeige

Anzeige

Netzkabel(Ca. 3m)

USB-Kabel(Ca. 3m)

Wechselstromadapter(Bilddiagramm)
FG-D-CWP / FG-D-ACWP gemeinsames Zubehör
Schnellstartanleitung

Gummiverpackung(Wägezellenkabel Durchmesser /Verwendung bei Φ3,5 bis 7,0)

15Gummifüße für Displayständer (4 Stk.)

DE

3. Vorsichtshinweise
3.1. Sicherheitsvorkehrungen zur Installation der -Anzeige
GEFAHR
Erden Sie die Waage, damit der Benutzer keinen elektrischen Schlag erleidet.
Um einen elektrischen Schock zu vermeiden, berühren Sie das USB-Kabel/Netzkabel nicht mit nassen
Händen.
Der USB-Stecker ist nicht wasserdicht. Verwenden Sie einen USB-Anschluss in einem Bereich, in dem
der Stecker nicht nass wird. (Für FG-D-CWP (USB-Strom-Modell))Der Netzstecker ist nicht wasserdicht. Verwenden Sie einen Netzanschluss in einem Bereich, indem der Stecker nicht nass wird. (Für FG-D-ACWP (Netzstrom-Modell))Stellen Sie die Waage nicht in einer Umgebung auf, in der entzündliche oder ätzende Gase auftreten.
Üben Sie keine übermäßige Kraft auf die Kabel aus.
Die Waage ist schwer. Seien Sie vorsichtig, da sie beim Anheben umfallen kann.
Verwenden Sie abgeschirmte Wägezellenkabel.
Bitte lesen Sie „Sicherheitsvorkehrungen zur Installation der Waage“ in [Bedienungshandbuch: FG-CWP /FG-D-ACWP Serie ] auf der A&D-Website (https://www.aandd.jp) für weitere Sicherheitsvorkehrungen,Vorsichtshinweise als montierte Waage, Lagerung und Reinigung.
Bitte lesen Sie „Installation und Vorsichtsmaßnahmen“ in [Bedienungshandbuch: FG-D-CWP / FG-DACWP] auf der A&D-Website (https://www.aandd.jp) zur Verbindung von Wägezellen, Installation vonAnzeigen, für Funktionseinstellungen und Empfindlichkeitseinstellungen usw.

3.2. Verbindung von Wägezellen
FG-D-CWP / FG-D-ACWP hat zwei Verbindungskonfigurationen: die 6-Draht-Konfiguration und die 4Draht- Konfiguration.
Wir empfehlen die Verbindung mit der 6-Draht-Konfiguration für optimale Genauigkeit und Stabilität.
Der zulässige Durchmesser des Wägezellenkabels beträgt von φ3,5 bis φ9,0 mm.
Die Länge des Wägezellenkabels soll weniger als 30 m betragen. Bei einer Konfiguration mit 4 Adern
empfehlen wir eine Länge von bis zu 5 m.
Anschluss Nr.
1
2
3
4
5
6
7

SIGSIG+
EXCSENSEN+
EXC+
ABSCHIRMUNG

Funktion
Wägezellen-Eingang (-)
Wägezellen-Eingang (+)
Wägezellen-Erregerspannung (-)
Sensoreingang (-)
Sensoreingang (+)
Wägezellen-Erregerspannung (+)
Abschirmung

16

DE
Wägezelle

Kabel

Kabel

(A) 6-Draht-Konfigurationsverbindung(B) 4-Draht-Konfigurationsverbindung

zur Wägezelle (empfohlen)

Verbindung zu Wägezelle

4. Teilenamen<FG-D-CWP (USB-Strom-Modell)>
Displayeinheit (SUS304)
Displayeinheitknopf
Seriennummer

Displayständer /Wandhaltebügel

Wägezellen-Panel
Wägezellen-Anschluss
USB-Kabel (Ca. 3m)
USB-Stecke<FG-D-ACWP (Netzstrom-Modell)>
Displayeinheit (SUS304)

Displayeinheitknopf
Seriennummer

Wägezellen-Panel

Displayständer /Wandhaltebügel

Wägezellen-Anschluss
Netzkabel (ca. 3 m)

Erd-Terminal
Erde

17

DE

4.1. Display und Symbole
Displayeinheit

Wäge-Modus
Vergleichs-LED
Modi fürjede Funktion

Gewichtseinheiten

Display/Symbol

Beschreibung
Diese Anzeige leuchtet, wenn der Gewichtswert stabil ist. Dadurch wird angezeigt,
dass die Waage im geeigneten Zustand zum Lesen von Gewichtswerten ist.
Diese Anzeige leuchtet, wenn das Gewicht des Containers (Eigengewicht)
abgezogen wird.
Diese Anzeige leuchtet, wenn die Waage sich im Nullpunkt (Referenzpunkt zum
Wiegen) befindet.

STABLENET
ZERO
Gewichtseinheiten

„

“, „

“, „

“, „

“,„

“ und „

“ leuchten.

Es leuchtet, wenn das Display gehalten wird.
Diese Funktion erkennt Stöße auf den Massensensor und zeigt den Stoßgrad an.
Leuchtet, wenn die Verbindung mit dem kabellosen Kommunikationsempfänger
hergestellt ist.
Während die Vergleichsfunktion verwendet wird, wird der Gewichtswert mit denvoreingestellten Schwellenwerten verglichen und die Anzeige zeigt das Gewicht an.
Wechselstrom
Erdungsanschluss
Einzelheiten zur Bedienmethode, Funktionen sowie eine Funktionstabelle finden Sie in[Bedienungshandbuch: FG-CWP / FG-ACWP-Serie]auf unserer Website (https://www.aandd.jp).
Bitte beachten Sie jedoch, dass Funktionen im Zusammenhang mit USB und mobilem Akku nicht
mit FG-D-ACWP (Netzstrom-Modell) verwendet werden können.

4.2. Bedienungstasten
Bedienungstaste

Beschreibung
ON/OFF-Taste
Das Display wird angezeigt oder ausgeblendet.
SAMPLE-Taste
Durch die Auswahl von „pcs“ als Einheit wird der individuelle
Gewichtseinstellungsmodus aktiviert.
Durch Gedrückthalten (für etwa 2 Sekunden) wird der VergleichsSchwellenwerteinstellmodus aktiviert.
18

DE
Bedienungstaste

Beschreibung
PRINT-Taste
Gibt den Gewichtswert als Daten aus.
Erhöht den Wert der Ziffer, die bei der Einstellung aufblinkt.
MODE-Taste
Wechselt die Gewichtseinheit.
Springt zur rechten Ziffer, die aufblinkt, wenn Einstellungen vorgenommen werden.
ZERO-Taste
Ausgehend von dem zum Zeitpunkt des Einschaltens der Anzeige gewogenen
Nullpunkt wird durch Drücken der Taste ZERO der Nullpunktanzeiger
eingeschaltet, sobald die Anzeige auf Null geht, wenn der Gewichtswert innerhalb
von ±2 % des Wägebereichs stabil ist.
TARE-Taste
Das Gewicht des Gegenstands auf der Waageschale wird als Gewicht eines
Containers (Eigengewicht) abgezogen.

5. Wartung･Siehe “3. Vorsichtshinweise” für Informationen zur Verwendung.
･Siehe “5.3 Fehlercode-Tabelle” und den entsprechenden Modus für den angezeigten Fehlercode.
･Siehe [Bedienungshandbuch FG-CWP / FG-ACWP -Serie]auf der A&D-Website (https://www.aandd.jp) zur Empfindlichkeitseinstellung.
･Prüfen Sie regelmäßig die Genauigkeit der Waage. Passen Sie die Waage an, wenn sie an einen anderen
Ort bewegt wurde oder sich die Umgebung verändert hat.

5.1. Reparatur
Nehmen Sie die Waage nicht auseinander und setzen Sie die Waage nicht zusammen ohne einen
autorisierten Service-Techniker. Dies kann zu einem elektrischen Schock oder Schäden an der Waage
usw. führen. In diesem Fall ist die Reparatur nicht von der Garantie abgedeckt. Wenden Sie sich an ihren
lokalen A&D-Händler, wenn Ihre Waage gewartet oder repariert werden muss.

5.2. Fehleranzeige
In dieser Situation
Bestätigen Sie folgendes
Nichts wird angezeigt.
･Hat der Hauptstrom die korrekte Spannung?
Die Waage lässt sich nicht･Ist das Hauptstromkabel ordnungsgemäß angeschlossen?
einschalten.
･Prüfen Sie die Umgebung der Waagschale.
･Befindet sich etwas auf der Waagschale?
Die Waage zeigt am Anfang･Siehe [Bedienungshandbuch FG-D-CWP / FG-D-ACWP-Serie] auf derkeine Null an.
A&D-Website (https://www.aandd.jp) und führen Sie die Einstellung der
Nullpunktempfindlichkeit durch.
wird angezeigt.

･Wäge-Fehler, der „Überladen“ bedeutet.

wird angezeigt.

･Zeigt an, dass das Gewicht der Probe zu leicht ist, um das
Einheitsgewicht im Zählmodus einzustellen.

wird angezeigt.

･Empfindlichkeitseinstellungsfehler, der bedeutet „zu leicht“.

wird angezeigt.

･Wird angezeigt, wenn der Hauptstrom ausfällt.

19

DE
In dieser Situation

Bestätigen Sie folgendes

wird
angezeigt undfährt nicht fort.

･Der gewogene Wert ist wegen Verschiebungen, Vibrationen oder anderen
Faktoren instabil.
Ein Windzug oder Vibrationen beeinflussen möglicherweise die Messung.
Prüfen Sie die Umgebung der Waagschale.
･Prüfen Sie die Verbindung des Wägezellenkabels.
･Keine Null-Anzeige, wenn das Display eingeschaltet wird.
Entfernen Sie jegliche Gegenstände von der Schale.
･Siehe [Bedienungshandbuch FG-D-CWP / FG-D-ACWP-Serie] auf der
A&D-Website (https://www.aandd.jp) und führen Sie die Einstellung der
Nullpunktempfindlichkeit durch.

Repariertes Display.

･Haben Sie die „Haltefunktion“ verwendet?
Siehe [Bedienungshandbuch FG-CWP / FG-ACWP-Serie] auf der A&DWebsite (https://www.aandd.jp) und ändern Sie die Funktion.
･Schalten Sie die Waage aus und wieder ein.

5.3. Fehlercode-Tabelle
Falls einer der folgenden Fehler angezeigt wird, versuchen Sie, das Display aus- und wieder einzuschalten.

Anzeige

Inhaltwird angezeigt

Der Temperatursensor ist fehlgeschlagen.

wird angezeigt

Der Speicher (Schaltkreis) ist fehlgeschlagen.

wird angezeigt

Der interne Schaltkreis ist fehlgeschlagen.

wird angezeigt

Der Massensensor ist fehlgeschlagen.

6. Spezifikationen
Die Spezifikationen können ohne vorherige Anmeldung verändert werden.
Eingangsempfindlichkeit
0,1 μV/d min. (d: Mindestteilung)Eingangssignalbereich-16 mV bis 16 mV
Wägezellen-Erregerspannung
5 V ± 10 %Antriebsleistung der WägezelleBis zu 4 x 350 ohm Wägezellen
Temperaturkoeffizient Null
±0,02 μV/°C typ. ± 0,1 μV/°C max.
Temperaturkoeffizient±3 ppm/°C typ. ±15 ppm/°C max.
MessspanneLinearität±0,01 % des Messbereichsendwerts
Maximale Displayauflösung
100000Anzeige
7-Segment-LED mit Hintergrundlicht (Buchstabengröße 30 mm)
Display-Update
Ca. 10-mal/Sekunde
Betriebsbedingungen-10 °C bis 40 °C, 85 % R.F. oder weniger (nicht kondensierend)
Installationsumgebung
Nur Innen benutzen
Höhe
Höhenlage bis 2.000 m<FG-D-CWP (USB-Strom-Modell)>
Stromversorgung
Wechselstromadapter, mobiler Akku oder USB Typ-A-Anschluss
USB-Kabellänge: Ca. 3 m(Das USB-Kabel wird auch zur Kommunikation verwendet.)<FG-D-ACWP (Netzstrom-Modell)>Netzstrom (100 V bis 240 V, 50/60 Hz, 0,1 A max.)
Netzkabellänge: Ca. 3 m
Überspannungskategorie *II
Verschmutzungsgrad *2
Maße
Breite 240 mm × Tiefe 76 mm × Höhe 155 mm
Gewicht (ungefähr)
1,6 kg (inklusive Ständer)
Staub- und Wasser- Schutz
Konform mit IP67

* Gilt nur für FG-D-ACWP (Wechselstrommodell).
20

ES
Guía de inicio rápido (ES)

1. Introducción
Esta guía de inicio rápido describe las funciones y operaciones básicas de este producto.
El indicador de pesaje FG-D-CWP / FG-D-ACWP se puede utilizar en una báscula de plataforma a prueba deagua con certificación IP67. Tiene especificaciones y funciones similares a un indicador de la serie FG-CWP.
Los siguientes manuales de instrucciones proporcionan información detallada sobre el FG-D-CWP / FGD-ACWP.
Dichos manuales se pueden descargar del sitio web de A&D (https://www.aandd.jp).
Escanee este código QR para obtener este manual.

[Manual de instrucciones del FG-D-CWP / FG-D-ACWP] Este manual de instrucciones se utiliza para
conectar e instalar el FG-D-CWP / FG-D-ACWP a una plataforma de pesaje (célula de carga).
[Manual de instrucciones de la serie FG-CWP / FG-ACWP]) Este es un manual para comprender yutilizar plenamente la báscula de plataforma digital a prueba de agua de la serie FG-CWP.

2. Elementos incluidos
El producto viene con los siguientes elementos.
FG-D-CWP (Modelo con alimentación por USB)

FG-D-ACWP (Modelo con alimentación de CA)

FG-D-CWP

FG-D-ACWP

Etiqueta del modelo(1083039230-1)

Escriba los valores
de la báscula, etc.

Montar aquíMontar aquí

Etiqueta del modelo(1083039230-3)

Escriba los valores
de la báscula, etc.

Indicator

Indicador

Cable USB
(3 m aprox.)

Cable de CA(3 m aprox.)

Adaptador de CA(Esquema de la imagen)Accesorios comunes para FG-D-CWP / FG-D-ACWP
Guía de inicio rápido

Embalaje de goma(cables de la célula decargadiámetro / Utilizado cuando
se usa un flujo magnético de 3,5
a 7,0 Φ)21

Patas de goma para el soporte de lapantalla (4 pzas.)

ES

3. Medidas de precaución
3.1. Medidas de precaución para la instalación del indicador
PELIGRO
Conecte la báscula a tierra para evitar que el usuario quede expuesto a una descarga eléctrica.
Para evitar descargas eléctricas, no maneje el cable USB / cable de CA con las manos mojadas.
El conector USB no es resistente al agua. Utilice una toma USB ubicada en un lugar en el que el conector
no se pueda mojar. (Para FG-D-CWP (modelo con alimentación por USB))
El conector de CA no es resistente al agua. Utilice una toma eléctrica ubicada en un lugar en elque el conector no se pueda mojar. (Para FG-D-ACWP (modelo con alimentación de CA))No instale la báscula donde haya gas inflamable o corrosivo.
No ejerza demasiada fuerza sobre los cables.
La báscula es pesada. Tenga cuidado, ya que si la levanta se puede caer.
Utilice cables apantallados para célula de carga.
Lea “Medidas de precaución para la instalación de la báscula” en [Manual de instrucciones: Serie FGCWP /FG-ACWP] en el sitio web de A&D (https://www.aandd.jp) pata obtener información sobre otras
medidas de precaución para el montaje de la báscula, el almacenamiento y la limpieza.
Lea “Instalación y medidas de precaución” en [Manual de instrucciones: FG-D-CWP / FG-D-ACWP] en elsitio web de A&D (https://www.aandd.jp) para obtener información sobre la conexión de las células decarga, la instalación de los indicadores, los ajustes de función, el ajuste de la sensibilidad, etc.

3.2. Conexión de la célula de carga
El FG-D-CWP / FG-D-ACWP cuenta con dos configuraciones de conexión: la configuración de 6 cables yla configuración de 4 cables.
Recomendamos la conexión utilizando la configuración de 6 cables para una precisión y estabilidadóptimas.
El diámetro aplicable del cable de la célula de carga es de entre 3,5 φ y 9,0 φ mm.
La longitud del cable de la célula de carga debería ser inferior a 30 m. Para la configuración de 4 cables
recomendamos utilizar uno de 5 m o menos.
N.º de terminal
1
2
3
4
5
6
7

SIGSIG+
EXCSENSEN+EXC+
SHIELD

Función
Entrada de la célula de carga (-)
Entrada de la célula de carga (+)
Voltaje de excitación de la célula de
Entrada de detección (-)
Entrada de detección (+)
Voltaje de excitación de la célula deProtector

22

ES
Célula de carga

Cable

Cable

Entrada de la célula de carga (-)

Entrada de detección (-)
Entrada de detección (+)
Protector

Protector(A) Conexión a la célula de cargacon configuración de conexión de 6 cables (recomendado)(B) Conexión a la célula de cargacon configuración de conexión de 4 cables

4. Nombre de las piezas<FG-D-CWP (Modelo con alimentación por USB)>
Unidad de la pantalla(SUS304)

Mando giratorio
de la pantalla
Número de serie

Soporte de la pantalla /Soporte para montaje
en pared

Panel de la célula
de carga
Conector de lacélula de carga
Cable USB (3 m aprox.)
Conector USB<FG-D-ACWP (Modelo con alimentación de CA)>
Mando giratorio
de la pantalla

Unidad de la pantalla(SUS304)

Número de serie

Soporte de la pantalla /Soporte para montaje
en pared

Panel de la célula
de carga
Conector de lacélula de carga
Cable de CA (aprox. 3 m)

Tierra23

terminal de tierra

ES

4.1. Pantalla y símbolos
Unidad de la pantalla
Estadodel pesaje
Comparador LED
Condiciones paracada función

Unidades de pesaje

Pantalla/símbolo
STABLENETZERO
Unidades de pesaje

Descripción
Se enciende cuando el valor de pesaje es estable e indica que la báscula estáen las condiciones adecuadas para la lectura de valores de pesaje.
Se enciende cuando se resta el peso del recipiente (tara).
Se enciende cuando la balanza está en el punto cero (punto de referencia parael pesaje).
“
”, “
”, “
”, “
”,“
” y “ ” están encendidos.
Se enciende cuando se sostiene la pantalla.
Esta función detecta el impacto en la sección del sensor de masa y mostrar el
nivel de impacto.
Se enciende cuando se establece la conexión con el receptor de comunicación
inalámbrica.
Cuando se usa la función de comparador, el valor de pesaje se compara
usando los valores umbrales preestablecidos y el indicador muestra el
resultado.
Corriente alterna.
Terminal de tierra.

Para más información sobre el método de operación, las distintas funciones, la tabla de funciones,
etc., consulte el [Manual de instrucciones: serie FG-CWP / FG-ACWP]
en nuestro sitio web (https://www.aandd.jp).
Sin embargo, tenga en cuenta que las funciones relacionadas con el USB y la batería móvil no sepueden utilizar con FG-D-ACWP (modelo con alimentación de CA).

4.2. Operaciones de las teclas
Operación de la
tecla

Descripción
Tecla ON/OFF
Muestra u oculta la pantalla.
Tecla SAMPLE
Al seleccionar “pcs” como la unidad, se activa el modo de ajuste de peso
individual.
Mantener pulsado (durante unos 2 segundos) activa el modo ajuste del umbraldel comparador.
24

ES
Operación de la
tecla

Descripción
Tecla PRINT
Envía el valor de pesaje como datos.
Aumenta el valor del dígito que parpadea durante el ajuste.
Tecla MODE
Cambia la unidad de pesaje.
Desplaza a la derecha el dígito que parpadea al realizar los ajustes.
Tecla ZERO
Según el punto cero pesado en el momento de encender la pantalla, al pulsar la
tecla ZERO cuando el valor de peso es estable dentro de ±2 % de la capacidad
de pesaje, el indicador de punto cero se enciende en cuanto la pantalla se ponea cero.
Tecla TAREEl peso del elemento en el plato de pesaje se resta como el peso de un recipiente(peso tara).

5. Mantenimiento･Consultar “3. Medidas de precaución” para obtener información sobre el uso.
･Consultar “5.3. Tabla de códigos de error” y el modo correspondiente para el código de error mostrado.
･Consulte el [Manual de instrucciones de la serie FG-CWP / FG-ACWP] en el sitio web A&D(https://www.aandd.jp) para obtener información sobre el ajuste de sensibilidad.
･Compruebe periódicamente la precisión del pesaje. Ajuste la báscula si se ha movido a otra ubicación o si
el entorno ha cambiado.

5.1. Reparación
No desmonte/monte la báscula sin la presencia de un técnico autorizado. Si lo hace, puede provocar una
descarga eléctrica, dañar la báscula, etc. En tal caso, la garantía no cubre la reparación. Póngase en
contacto con su distribuidor local de A&D si su báscula necesita mantenimiento o reparación.

5.2. Visualización de errores
En esta situación

Confirme estos elementos

No se muestra nada.
La báscula no se enciende.

･¿El voltaje de la alimentación principal es el correcto?
･¿Está conectado correctamente el cable de alimentación principal?
･Compruebe el entorno del plato de pesaje.
･¿Hay algo en el plato de pesaje?
･Consulte el [Manual de instrucciones del FG-D-CWP / FG-D-ACWP] en elsitio web de A&D (https://www.aandd.jp) y realice el ajuste de sensibilidad
del punto cero.
･Error de pesaje que significa “Sobrecarga”.
･Indica que el peso de la muestra es demasiado ligero para establecer elpeso de la unidad en el modo de recuento.
･Error de ajuste de sensibilidad que significa “Demasiado baja”.

La báscula no muestra ceroal inicio.
Se muestra
Se muestra
Se muestra
Se muestra

Se muestra
y no continúa.

･Aparece cuando la alimentación principal se corta.
･El valor pesado es inestable debido a la oscilación, la vibración y otros
factores.
Una corriente de aire o una vibración pueden estar afectando a la
medición.
Compruebe el entorno del plato de pesaje.
･Compruebe la conexión del cable de la célula de carga.
･No se muestra cero cuando la pantalla se enciende.
Retire cualquier elemento del plato.
･Consulte el [Manual de instrucciones del FG-D-CWP] en el sitio web de A&D(https://www.aandd.jp) y realice el ajuste de sensibilidad del punto cero.

25

ES
En esta situación
Pantalla fija.

Confirme estos elementos･¿Ha utilizado la “función hold”?
Consulte el [Manual de instrucciones de la serie FG-CWP / FG-ACWP] enel sitio web A&D (https://www.aandd.jp) y cambie la función.
･Apague la báscula y vuelva a encenderla.

5.3. Tabla de códigos de error
Cuando se muestra alguno de los siguientes códigos de error, pruebe apagar la pantalla y volver aencenderla.

PantallaContenido

Se muestra

Fallo en el sensor de temperatura.

Se muestra

Fallo en la memoria (circuito).

Se muestra

Fallo en el sistema interno de circuitos.

Se muestra

Fallo en el sensor de masa.

6. Especificaciones
Las especificaciones están sujetas a cambio sin previo aviso.
Sensibilidad de entrada
Intervalo de la señal deentradaVoltaje de excitación de lacélula de carga
Capacidad de resistencia dela célula de carga
Coeficiente de temperatura
ceroIntervalo de coeficiente detemperatura
Linealidad
Valor de visualización máximo
Pantalla
Actualización de pantalla
Condiciones de operación
Entorno de instalación
Altitud
Alimentación

Categoría de sobrevoltaje *Grado de contaminación *Dimensiones
Peso (aproximadamente)
Protección contra el polvo y el
agua

0.1 μV/d min. (d: división mínima)
De -16 mV a 16 mV
5 V ± 10 %Hasta 4 células de carga de 350 ohmios±0.02 μV/°C típ. ±0.1 μV/°C máx.
±3 ppm/°C típ. ±15 ppm/°C máx.
±0.01 % del fondo de escala
100000LCD de 7 segmentos con retroiluminación (altura del carácter 30 mm)
10 veces/segundo aprox.
De -10 °C a 40 °C, 85 % de humedad relativa o menos (sincondensación)
Solamente para uso en interiores
Altitud hasta 2.000 m<FG-D-CWP (modelo con alimentación por USB)>
Alimentación suministrada desde el adaptador de CA, la batería móvil oel puerto USB tipo A
Longitud del cable USB: 3 m aprox.
(El cable USB también se utiliza para la comunicación).
<FG-D-ACWP (modelo con alimentación por AC)>Red de CA (100 V a 240 V, 50/60 Hz, 0,1 A máx.)Longitud del cable de CA: 3 m aprox.
II
2
Ancho 240 mm × Profundidad 76 mm × Altura 155 mm
1,6 kg (incluyendo el soporte)
Conforme con IP67

* Aplicable únicamente a FG-D-ACWP (modelo de alimentación de CA)

26

IT
Guida introduttiva (IT)

1. Introduzione
Questa guida introduttiva descrive le funzioni di base e le operazioni di questo prodotto.
È possibile usare l’indicatore di pesatura FG-D-CWP / FG-D-ACWP su una bilancia a piattaforma
impermeabile con prestazioni IP67. Ha specifiche tecniche e funzioni simili a quelle di un indicatore dellaserie FG-CWP.
I seguenti manuali di istruzioni includono informazioni dettagliate su FG-D-CWP / FG-D-ACWP.
Questi manuali possono essere scaricati dal sito web di A&D (https://www.aandd.jp).
Eseguire la scansione di questo codice QR per questo manuale.

[Manuale di istruzioni di FG-D-CWP / FG-D-ACWP]
Manuale di istruzioni utilizzato per il collegamento e l’installazione di FG-D-CWP / FG-D-ACWP su una
piattaforma di pesatura (cella di carico).
[Manuale di istruzioni della serie FG-CWP / FG-ACWP]
Manuale di istruzioni per comprendere e utilizzare in modo completo la bilancia digitale a piattaforma
impermeabile della serie FG-CWP.

2. Articoli inclusi
I seguenti articoli sono forniti in dotazione con il prodotto.
FG-D-CWP (Modello ad alimentazione USB)

FG-D-ACWP (Modello ad alimentazione CA)

FG-D-CWP

FG-D-ACWP

Etichetta del modello(1083039230-1)

Attaccarla qui

Scrivere la potenzadella bilancia, ecc.

Etichetta del modello(1083039230-3)

Attaccarla qui

Indicatore

Scrivere la potenzadella bilancia, ecc.

Indicatore

Cavo USB(circa 3 m)

Cavo CA(circa 3 m)

Adattatore CA(Illustrazione)
Accessori comuni per FG-D-CWP / FG-D-ACWP
Guida introduttiva

Guarnizione in gomma(Diametro dei cavi della cella dicarico/Utilizzata con Φ3,5-7,0)27

Piedini in gomma per il supporto deldisplay (4 pz.)

IT

3. Precauzioni
3.1. Precauzioni per l’installazione dell’indicatore
PERICOLO
Mettere a terra la bilancia per evitare che l’utente possa essere esposto a scosse elettriche.
Per evitare le scosse elettriche, non maneggiare il cavo USB / cavo CA con le mani bagnate.
La spina USB non è resistente all’acqua. Utilizzare una presa USB situata in un luogo in cui la spina nonsi bagni. (Per FG-D-CWP (modello ad alimentazione USB))
La spina CA non è resistente all’acqua. Utilizzare una presa elettrica situata in un luogo in cui laspina non si bagni. (Per FG-D-ACWP (modello ad alimentazione CA))Non installare la bilancia in presenza di gas infiammabili o corrosivi.
Non esercitare forza eccessiva sui cavi.
La bilancia è pesante. Maneggiarla con cura nel sollevarla, perché potrebbe ribaltarsi.
Utilizzare cavi per celle di carico schermati.
Leggere “Precauzioni per l’installazione della bilancia” nel [Manuale di istruzioni: serie FG-CWP / FG-ACWP]
sul sito web di A&D (https://www.aandd.jp) per altre precauzioni, utilizzo come bilancia assemblata,conservazione e pulizia.
Leggere “Installazione e precauzioni” nel [Manuale di istruzioni: FG-D-CWP / FG-D-ACWP] sul sito web di
A&D (https://www.aandd.jp) in relazione al collegamento della cella di carico, l’installazione di indicatori,
l’impostazione delle funzioni, la regolazione della sensibilità, ecc.

3.2. Collegamento della cella di carico
FG-D-CWP / FG-D-ACWP ha due configurazioni di collegamento: la configurazione a 6 cavi e laconfigurazione a 4
cavi.
Consigliamo di effettuare il collegamento con una configurazione a 6 cavi per la massima precisione e
stabilità.
Il diametro idoneo del cavo della cella di carico va da φ3,5 a φ9,0 mm.
La lunghezza del cavo della cella di carico deve essere inferiore a 30 m. Per la configurazione a 4 cavi, si
consiglia di utilizzare una lunghezza di 5 m o inferiore.
N. del terminale
1
2
3
4
5
6
7

SIGSIG+
EXCSENSEN+EXC+
SHIELD

Funzione
Ingresso della cella di carico (-)
Ingresso della cella di carico (+)
Tensione di eccitazione della cella di carico (-)
Ingresso del rilevamento (-)
Ingresso del rilevamento (+)
Tensione di eccitazione della cella di carico (+)
Schermo28

IT
Cella di carico

Cavo

Cavo

Ingresso della cella di carico (-)

Ingresso del rilevamento (-)
Ingresso del rilevamento (+)
SchermoSchermo(A) Collegamento alla cella di caricocon configurazione a 6 cavi (consigliata)(B) Collegamento alla cella di caricocon configurazione a 4 cavi

4. Nomi dei componenti<FG-D-CWP (Modello ad alimentazione USB)>
Display (SUS304)
Manopola del display
Numero di serie

Pannello dellacella di carico
Connettore dellacella di carico

Supporto del display /Staffa per il montaggio
a parete

Cavo USB (circa 3 m)
Spina USB

<FG-D-ACWP (Modelo con alimentación de CA)>
Display (SUS304)

Manopola del display
Numero di serie

Supporto del display /Staffa per il montaggio
a parete

Pannello dellacella di carico
Connettore dellacella di carico
Cavo CA (circa 3 m)

Terminale di terra
Terra

29

IT

4.1. Display e simboli
Display
Condizione
di pesatura
Comparatore a LED
Condizioni perciascuna funzione

Display/simbolo
STABLENET
ZERO
Unità di pesatura

Unità di pesatura

Descrizione
Si illumina quando il valore di pesatura è stabile, indicando che la bilancia è nella
condizione giusta per la lettura dei valori della pesatura.
Si illumina quando si sottrae il peso del contenitore (tara).
Si illumina quando la bilancia è sul punto zero (punto di riferimento per lapesatura).
Si illuminano “

”, “

”, “

”, “

”, “

”e“

”.

Si accende quando il display è bloccato.
Questa funzione rileva l’impatto sulla sezione del sensore di massa e visualizza il
livello di impatto.
Si accende quando si stabilisce la connessione con il ricevitore di comunicazione
wireless.
Durante l’utilizzo della funzione del comparatore, il valore di pesatura viene
confrontato utilizzando i valori soglia preimpostati e l’indicatore visualizza ilrisultato.
Corrente alternata.
Terminale di terra.
Per informazioni dettagliate sul metodo di funzionamento, sulle varie funzioni, sulla tabella delle
funzioni, ecc. fare riferimento al [Manuale di istruzioni: serie FG-CWP / FG-ACWP]
sul nostro sito web (https://www.aandd.jp).
Tuttavia, si noti che le funzioni relative a USB e batteria mobile non possono essere utilizzate con
FG-D-ACWP (modello ad alimentazione CA).

4.2. Tasti funzione
Tasto funzione

Descrizione
Tasto ON/OFF
Mostra o nasconde il display.
Tasto SAMPLE
Se si seleziona “pcs” come unità si attiva la modalità di impostazione del peso
singolo.
Tenendolo premuto (per circa 2 secondi) si attiva la modalità di impostazione dellasoglia del comparatore.

30

IT
Tasto funzione

Descrizione
Tasto PRINT
Emette il valore di pesatura sotto forma di dati.
Aumenta il valore della cifra che lampeggia durante l’impostazione.
Tasto MODE
Commuta l’unità di pesatura.
Sposta a destra la cifra che lampeggia durante le impostazioni.
Tasto ZERO
A seconda del punto zero pesato al momento dell’accensione del display, se si
preme il tasto ZERO quando il valore di peso è stabile entro il ±2% della capacità dipesatura, l’indicatore del punto zero si accende non appena il display diventa zero.
Tasto TARE
Il peso dell’articolo sul piatto per la pesatura viene sottratto come peso delcontenitore (peso della tara).

5. Manutenzione･Fare riferimento a “3. Precauzioni” in relazione all’uso.
･Fare riferimento a “5.3. Tabella dei codici errore” e alla modalità corrispondente per il codice errore visualizzato.
･Fare riferimento al [Manuale di istruzioni della serie FG-CWP / FG-ACWP] sul sito web di A&D(https://www.aandd.jp) per la regolazione della sensibilità.
･Controllare periodicamente la precisione della pesatura. Se la bilancia è stata spostata in un luogo diverso
o se l’ambiente è cambiato, regolarla.

5.1. Riparazione
Non smontare/montare la bilancia se non in presenza di un tecnico autorizzato. In caso contrario, si
potrebbero causare scosse elettriche o danni alla bilancia, ecc. In tal caso, la riparazione non sarà copertada garanzia. Per esigenze di assistenza o riparazione della bilancia, contattare il rivenditore locale di A&D.

5.2. Visualizzazione degli errori
In questa situazione

Verificare questo

Non appare nulla sul display.
La bilancia non si accende.

･La tensione dell’alimentazione principale è quella corretta?
･Il cavo di alimentazione principale è collegato correttamente?

Sulla bilancia non appare lo
zero all’inizio.

･Controllare intorno al piatto per la pesatura.
･C’è qualcosa sul piatto per la pesatura?
･Fare riferimento al [Manuale di istruzioni di FG-D-CWP / FG-D-ACWP] sul
sito web di A&D (https://www.aandd.jp) ed eseguire la regolazione dellasensibilità del punto zero.

Si visualizza.

･Errore di pesatura che significa “Sovraccarico”.

Si visualizza.

･Indica che il peso del campione è troppo leggero per poter impostare ilpeso dell’unità nella modalità di conteggio.

Si visualizza.

･Errore di regolazione della sensibilità che significa “Troppo leggero”.

Si visualizza.

･Appare in caso di caduta dell’alimentazione principale.

Si visualizza
e nonavanza.

･Il valore pesato è instabile a causa di deriva, vibrazione o altri fattori.
Potrebbe esserci una vibrazione o una folata d’aria che interferisce con la
misurazione.
Controllare intorno al piatto per la pesatura.
･Controllare il collegamento del cavo della cella di carico.
･Non si visualizza lo zero quando si accende il display.
Rimuovere qualunque oggetto dal piatto.
･Fare riferimento al [Manuale di istruzioni di FG-D-CWP / FG-D-ACWP] sul
sito web di A&D (https://www.aandd.jp) ed eseguire la regolazione dellasensibilità del punto zero.
31

IT
In questa situazione

Display fisso.

Verificare questo･È stata utilizzata la “funzione Hold”?
Fare riferimento al [Manuale di istruzioni della serie FG-CWP / FG-ACWP]
sul sito web di A&D (https://www.aandd.jp) e cambiare la funzione.
･Spegnere la bilancia e riaccenderla.

5.3. Tabella dei codici errore
Se si visualizza uno dei seguenti errori, provare a spegnere e riaccendere il display.

Display

Contenuto

Si visualizza

Guasto del sensore della temperatura.

Si visualizza

Guasto della memoria (circuito).

Si visualizza

Guasto del circuito interno.

Si visualizza

Guasto del sensore di massa.

6. Specifiche tecniche
Le specifiche tecniche sono soggette a modifiche senza preavviso.
Sensibilità di ingresso
0,1 μV/d min. (d: divisione minima)
Gamma del segnale diDa -16 mV a 16 mV
ingresso
Tensione di eccitazione della
5 V ±10%cella di carico
Capacità della cella di caricoFino a 4 celle di carico da 350 ohm
Zero del coefficiente di±0,02 μV/°C tip. ±0,1 μV/°C max.
temperatura
Gamma del coefficiente di±3 ppm/°C tip. ±15 ppm/°C max.
temperatura
Linearità±0,01% del fondo scala
Risoluzione massima del100.000
display
Display
LCD a 7 segmenti con retroilluminazione (altezza del carattere 30 mm)
Aggiornamento del display
Circa 10 volte al secondo
Condizioni operativeDa -10 °C a 40 °C, max. 85% di umidità relativa (senza condensa)
Ambiente di installazione
Solo per uso interno
Altitudine
Altitudine fino a 2.000 m
Alimentazione<FG-D-CWP (modello ad alimentazione USB)>
Alimentazione tramite adattatore CA, batteria mobile o porta USB Tipo-A
Lunghezza del cavo USB: circa 3 m(Il cavo USB è utilizzato anche per le comunicazioni.)<FG-D-ACWP (modello ad alimentazione CA)>
Rete CA (Da 100 V a 240 V, 50/60 Hz, 0,1 A massimo)
Lunghezza del cavo CA: circa 3 m
Categoria di sovratensione *II
Grado di inquinamento *2
Dimensioni
Larghezza 240 mm × Profondità 76 mm × Altezza 155 mm
Peso (approssimativo)
1,6 kg (supporto incluso)
Protezione da polvere e acqua Conforme a IP67

* Applicabile solo a FG-D-ACWP (modello con alimentazione CA).

32

CT
快速入門指南(CT)1. 簡介
此快速入門指南描述本產品的基本功能和操作。
FG-D-CWP / FG-D-ACWP 稱重指示器可用於防水等級為 IP67 的防水台秤。其規格和功能與 FG-CWP 系列
的指示器類似。
有關 FG-D-CWP / FG-D-ACWP 的詳細資訊，請參閱下列說明手冊。
這些手冊可從 A&D 網站 (https://www.aandd.jp) 下載。
掃描此二維碼即可取得本手冊。

[FG-D-CWP / FG-D-ACWP 說明手冊]
本說明手冊係於將 FG-D-CWP / FG-D-ACWP 連接並安裝至稱重台（荷重元）時使用。
[FG-CWP / FG-ACWP 系列說明手冊]
本說明手冊係用於認識和充分運用防水數位台秤 FG-CWP 系列。

2. 內容物產品的包裝內容物如下。
FG-D-CWP (USB 電源機型)
FG-D-CWP

貼在此處FG-D-ACWP (AC 電源機型)
FG-D-ACWP

型號標籤
(1083039230-1)寫下台秤的額定值等

貼在此處

指示器

型號標籤
(1083039230-3)寫下台秤的額定值等

指示器

USB 纜線（約 3 m）

AC 纜線（約 3 m）

變壓器（示意圖）

FG-D-CWP / FG-D-ACWP 通用配件
快速入門指南橡皮墊圈

顯示支架的橡膠支腳（4 件）（荷重元纜線直徑 / 使用 Φ3.5
至 7.0 時使用）

33

CT

3. 警告
3.1. 指示器安裝注意事項
危險將台秤接地，以免使用者觸電。
為了避免觸電，手潮溼時，請勿拿取 USB 纜線／AC 纜線。
USB 插頭無防水效果。使用插頭不會受潮的 USB 插座。（適用於 FG-D-CWP（USB 電源機型）
AC 插頭無防水效果。使用插頭不會受潮的電氣插座。（適用於 FG-D-ACWP（AC 電源機型）
請勿在有易燃氣體或腐蝕性氣體的地方放置台秤。
請勿用力拉扯纜線。
台秤很重。請留意抬起台秤時，台秤可能會翻落。
請使用屏蔽荷重元纜線。
請前往A&D 網站 (https://www.aandd.jp)，詳閱 [說明手冊：FG-CWP / FG-ACWP 系列]中的「台秤安裝注意
事項」，瞭解其他注意事項，以及台秤組裝、存放和清潔須知。
請前往 A&D 網站 (https://www.aandd.jp)，詳閱 [說明手冊：FG-D-CWP / FG-D-ACWP]中的「安裝和注意事項」，瞭解荷重元連接、指示器安裝、功能設定和靈敏度調整等資訊。

3.2. 荷重元連接
FG-D-CWP / FG-D-ACWP 具備兩種連接配置：6 線配置和 4 線配置。
建議使用 6 線配置進行連接，以獲得最佳準確性和穩定性。
荷重元纜線的適用直徑為 φ3.5 至 φ9.0 mm。
荷重元纜線的長度應小於 30 m。若為 4 線配置，建議使用 5 m 以下的纜線。
端子編號
1
2
3
4
5
6
7

SIGSIG+
EXCSENSEN+
EXC+
屏蔽

功能
荷重元輸入 (-)
荷重元輸入 (+)
荷重元激發電壓 (-)
感測輸入 (-)
感測輸入 (+)
荷重元激發電壓 (+)
屏蔽

34

CT
荷重元

纜線

纜線

荷重元輸入 (-)
荷重元輸入 (+)
荷重元激發電壓 (-)
感測輸入 (-)
感測輸入 (+)
荷重元激發電壓 (+)
屏蔽

屏蔽(A) 6 線配置的荷重元連接（建議）(B) 4 線配置的荷重元連接

4. 零件名稱<FG-D-CWP (USB 電源機型)>
顯示元件 (SUS304)
顯示元件旋鈕
序號

顯示支架／
壁掛式支架

荷重元面板
荷重元接頭
USB 纜線（約 3m）
USB 插頭<FG-D-ACWP (AC 電源機型)>
顯示元件 (SUS304)

顯示元件旋鈕
序號

荷重元面板

顯示支架／
壁掛式支架

荷重元接頭
接地端子

AC 纜線（約 3 m）
接地
35

CT

4.1. 顯示和符號
顯示元件
稱重情況
比較器 LED
各功能運作情況

稱重單位

顯示／符號

描述

穩定

稱重值趨於穩定時亮起，代表台秤處於正確狀態，可讀取稱重值。

淨重
歸零

減去容器重量（皮重）時亮起。
台秤在歸零點（稱重參考點）時亮起。

稱重單位

「

」、「 」、「」、「」、「」和「 」亮起。

保留顯示內容時亮起。
此功能用於偵測質量感測器部分受到的衝擊，並顯示衝擊等級。
與無線通訊接收器建立連線時亮起。

使用比較器功能時，將稱重值與預設閾值進行比較，且指示器顯示結果。

交流電
接地端子
如需操作方式、各種功能和功能表等詳細說明，請前往我們的網站 (https://www.aandd.jp)，參閱 [說明手冊：FG-CWP / FG-ACWP 系列]。
然而，請注意，與 USB 和行動電池相關的功能無法搭配 FG-D-ACWP（AC 電源機型）使用。

4.2. 操作按鍵
操作按鍵

描述
ON/OFF 鍵顯示或隱藏顯示內容。
SAMPLE 鍵選擇「pcs」作為單位後，啟動個別重量設定模式。
按住按鍵約 2 秒後，啟動比較器閾值設定模式。
PRINT 鍵將稱重值輸出為資料。
進行設定時，增加閃爍數字的數值。
MODE 鍵切換稱重單位。
進行設定時，切換至閃爍數字的右側。

36

CT
操作按鍵

描述
ZERO 鍵根據開啟顯示器時測得的歸零點，在重量值穩定介於 ±2 % 稱重上限之間時按下ZERO 鍵，一旦顯示數值歸零，歸零點指示燈就會立即亮起。
TARE 鍵為減去秤盤上物品的重量，即容器重量（皮重）。

5. 維護･有關使用方式，請參閱「3. 警告」。
･有關顯示的錯誤代碼，請參閱「5.3. 錯誤代碼表」和對應模式。
･有關靈敏度調整，請前往 A&D 網站 (https://www.aandd.jp)，參閱 [FG-CWP / FG-ACWP 系列說明手冊]。
･請定期檢查稱重準確性。如果將台秤移至其他位置或環境有所改變，請調整台秤。

5.1. 維修
請勿在沒有授權檢修工程師的情況下，拆解／組裝台秤，否則可能會造成觸電、台秤損壞等問題。如果因此需要維修，將不適用保固。如果您的台秤需要檢修或維修，請聯絡當地 A&D 經銷商。

5.2. 錯誤顯示
實際情況
未顯示任何內容。
台秤未開啟。

台秤一開始未歸零。

確認項目･主電源電壓是否正確？
･是否正確連接主電源線？
･請檢查秤盤。
･秤盤上是否有任何物品？
･請前往 A&D 網站 (https://www.aandd.jp)，參閱 [FG-D-CWP / FG-D-ACWP
說明手冊]，然後執行歸零點靈敏度調整。

顯示。

･代表「超重」的稱重錯誤。

顯示。

･代表樣品重量太輕，無法在計數模式下設定單位重量。

顯示。

･代表「太輕」的靈敏度調整錯誤。

顯示。

･主電源電量低下時顯示。

顯示，
且無法繼續操作。

･因偏移、震動等因素，導致稱重值無法趨於穩定。
微風或震動可能會影響測量結果。
請檢查秤盤。
･請檢查荷重元纜線的連接情形。
･開啟顯示器時，未顯示零。
請移除秤盤上的物品。
･請前往 A&D 網站 (https://www.aandd.jp)，參閱 [FG-D-CWP / FG-D-ACWP
說明手冊]，然後執行歸零點靈敏度調整。

顯示內容固定。

･是否使用「保留功能」？
請前往 A&D 網站 (https://www.aandd.jp)，參閱 [FG-CWP / FG-ACWP 系列說
明手冊]，然後變更功能。
･將台秤關閉再開啟。
37

CT

5.3. 錯誤代碼表顯示下列任一錯誤時，請試著將顯示器關閉再開啟。

內容

顯示
顯示

溫度感測器故障。

顯示

記憶體（迴路）故障。

顯示

內部電路故障。

顯示

質量感測器故障。

6. 規格
規格如有變動，恕不另行通知。
輸入靈敏度

0.1 μV/d min.（d：分度值）

輸入訊號範圍

-16 mV 至 16 mV

荷重元激發電壓5 V ± 10 %

荷重元磁碟容量

最多 4 x 350 ohm 荷重元溫度係數 Zero±0.02 μV/°C（標準值） ±0.1 μV/°C（最大值）溫度係數 Span±3 ppm/°C（標準值） ±15 ppm/°C（最大值）線性±0.01 % 滿刻度

最大顯示解析度

100000顯示

7 區段背光 LCD（字元高度 30 mm）

顯示更新

約每秒 10 次

操作條件-10 °C 至 40 °C，85 % R.H. 以下（非冷凝）安裝環境

僅限室內使用高度

海拔高度可達 2,000 米<FG-D-CWP（USB 電源機型）>
由 AC 變壓器、行動電池或 USB Type-A 連接埠供電
USB 纜線長度：約 3 m

電源（USB 纜線也可作為通訊用途。）<FG-D-ACWP（AC 電源機型）>
AC 電源（100V 至 240V，50/60Hz，最大 0.1A）
AC 纜線長度：約 3 m
過壓類別 *

II

污染程度*

2

尺寸

寬 240 mm × 深 76 mm × 高 155 mm

重量（約略值）1.6 kg（含支架）

防水防塵性能符合 IP67* 僅適用於 FG-D-ACWP（交流電源型號）38

THIS PAGE INTENTIONALLY LEFT BLANK.

A&D Company, Limited

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566

A&D ENGINEERING, INC.

47747 Warm Springs Blvd, Fremont, California 94539, U.S.A.
Tel: [1] (800) 726-3364
Weighing Support:[1] (888) 726-5931

A&D INSTRUMENTS LIMITED

Inspection Support:[1] (855) 332-8815

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

OX14 1DY

United Kingdom

A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264

Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3, корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Instruments India Private Limited

D-48( D-48, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599

A&D SCIENTECH TAIWAN LIMITED.

A&D台灣分公司 艾安得股份有限公司

台湾台北市中山區南京東路 2 段 206 號 11 樓之 2
( 11F-2, No.206, Sec.2, Nanjing E.Rd., Zhongshan Dist., Taipei City 10489, Taiwan, R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794

A&D INSTRUMENTS (THAILAND) LIMITEDบริษัท เอ แอนด์ ดี อินสทรูเม้นท์ (ไทยแลนด์) จำกัด
168/16 หมู่ที่ 1 ตำบลรังสิต อำเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย

( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911