Smart way to a healthy life0123

http://www.aandd.jp
http://www.wellnessconnected.jp/english/

Digital Blood Pressure Monitor

Precision Health Scale

UA-651BLE

UC-352BLE

Clinically validated for accuracy (ESH2010)
Bluetooth Ver.4.0LE
Bluetooth communication (Windows, iOS, Android)Internal memory of 30 data records
Irregular HeartBeat (IHB) indicator - LCD only
Extra large 3-line display for easier reading of results
One-touch measurement
Latex-free and metal-free, slimFit Cuff
Compact design for portability

Bluetooth Ver.4.0LE
Bluetooth communication (Windows, iOS, Android)
Durable glass plate scale
kg/lb selection
Internal memory of 90 data records
0 to 100kg : 100g resolution / 100 to 200kg : 200g resolution
0 to 220.0lb : 0.2lb resolution / 220.0 to 450.0lb : 0.5lb resolution
Simply step on start weighing
Beeps on completion of weighing
Low profile stylish design

Specifications
UA-651BLE

UC-352BLE

Measurement method
Display typeMeasurement range

Measurement accuracy
Power supply
Number of measurements

Memory storage
Cuff
Operating conditions
Transport/Storage conditions
Wireless communication
Communication specifications
Dimensions
WeightOptions

Oscillometric measurement
Digital LCD, Pressure/Pulse displayed simultaneouslyPressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse: 40 - 180 beats / minutePressure: ±3 mmHg
Pulse: ±5%
4 x 1.5V batteries (LR6, R6P or AA) orAC adapter (TB-233) (Not included)
Approx. 700 times LR6 (alkaline batteries)
Approx. 200 times R6P (manganese batteries)
With pressure value of 180 mmHg at room
temperature of 23 °C
30
SlimFit Cuff with plastic D-ring / Arm circumference : 22-32cm+10°C to +40°C / 15%RH to 85%RH/800 hPa to1060 hPa-20°C to +60°C / 10%RH to 95%RH
Bluetooth Ver.4.0LE BLP*Continua Design Guideline 2013 Version 4.1(ITU-T H.810 (12/2013) Part 0, Part 1)
Approx. 96 [W] x 68 [H] x 130 [D] mm
Approx. 250 g, excluding batteries
SlimFit Large Cuff (31-45cm) : CUF-F-LA
Wide Range Cuff (22-42cm) : CUF-I
AC Adapter (C/BF plug) : TB-233C/BF

A&D Connect

Maximum capacity
Display type
Minimum display
Setting items
Power source
Battery life
Memory storage
Operating conditions
Transport/Storage conditions
Wireless communication
Communication specifications
Dimensions
Weight200 kg / 450 lb
Digital LCD, Weight displayed after step off
0.1 kg (0 to 100 kg) / 0.2 kg (100 to 200 kg)
0.2 lb (0 to 220 lb) / 0.5 lb (220 to 450 lb)
Units: kg or lb
4 x 1.5V alkaline batteries (LR03 or A A A)
Approximately 1 year
(two measurements per day when alkaline batteries are used)
90+10°C to +40°C / 30%RH to 85%RH-10°C to +60°C / 30%RH to 85%RH
Bluetooth Ver.4.0LE WSP*Continua Design Guideline 2014 Version 5.0
Approx. 300 [W] x 300[D] x 31[H] mm
Approx. 2.1 kg (4.7lb)including batteries*Continua certification is instituted and promoted by PCHA to certify compliance to Continua DesignGuidelines. The guidelines were developed to realize an end-to-end interconnected service for healthcare
products.
The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any use ofsuch marks by A&D is under license. Other trademarks and trade names are those of their respective owners.
WellnessConnected™ is trademark of A&D Company, Limited or A&D Medical. Apple name and logo areregistered trademarks of Apple, Inc. Google, Android, Google Logo and Google Play are registeredtrademarks of Google, Inc. Facebook name and logo are trademarks of Facebook Inc. Twitter name and logo
are trademarks of Twitter Inc. Microsoft HealthVault icon is a registered logo of Microsoft Corporation in theUnited States and/or other countries.
A&D Medical is preparing a range of healthcare devices compatible with A&D Connect app.

Easy-to-use app, a solution for better accuracy
Monitor your blood pressure and track your weight and BMI trends – all with one app from a trusted
world-wide leader in connected health.A&D Connect guides you through the easy device set-up process
and provides at-a-glance results, goals, and trends for you and sharing data with your family and doctors.

Available on iOS 6.0 and above
iPhone 4S or newer, iPad Air, iPad mini

Dashboard

ListGraphSharing

Requires Android OS version 4.4and above
For compatible smartphones, pleasecheck WellnessConnected™ website

A&D INSTRUMENTS INDIA PRIVATE LIMITED509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India
Telephone : [91[(124)471-5555 Fax: [91](124)471-5599

BLE series-ADCC-01-IB-152