Wrist Digital Blood Pressure Monitor

Model UB-512
Instruction Manual
Original

Manuel d’instructions
Traduction

Manual de Instrucciones
Traducción

Manuale di Istruzioni
Traduzione

使用手冊
1WMPD4000980C

翻譯Contents
Dear Customers............................................................................................................................... 2
Preliminary Remarks ........................................................................................................................ 2
Precautions...................................................................................................................................... 2
Contraindications....................................................................................................................... 3
Parts Identification........................................................................................................................... 4Symbols........................................................................................................................................... 5
Using the Monitor ............................................................................................................................ 7
1. Installing / Changing the Batteries ......................................................................................... 7
2. Fastening the Cuff.................................................................................................................. 7
3. How to Take Accurate Measurements ..................................................................................... 8
4. Measurement (Operation) ....................................................................................................... 85. After Measurement................................................................................................................. 8Measurements ................................................................................................................................. 9
Normal Measurement with Storing Data...................................................................................... 9Measurements ......................................................................................................................... 10
Notes for Accurate Measurement.............................................................................................. 10Operation Mode ....................................................................................................................... 10
What is an Irregular Heartbeat? ...................................................................................................... 11
Pressure Bar Indicator .................................................................................................................... 11
WHO Classification Indicator .......................................................................................................... 12
About Blood Pressure..................................................................................................................... 12
What is Blood Pressure?............................................................................................................ 12
What is Hypertension and How is it Controlled? ........................................................................ 12
Why Measure Blood Pressure at Home? ..................................................................................... 13
WHO Blood Pressure Classification ........................................................................................... 13
Blood Pressure Variations ......................................................................................................... 13
Troubleshooting ............................................................................................................................ 14
Maintenance .................................................................................................................................. 15
Technical Data ............................................................................................................................... 15

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure monitor, one of the most advanced monitors
available today. Designed for ease of use and accuracy, this device will facilitate your daily blood pressure regimen.
We recommend that you read through this manual carefully before using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical Products. This is made evident by themark of conformity accompanied by the reference number of the designated authority.
The device is designed for use on adults only, not newborns or infants.
Environment for use.
The device is for use in the home healthcare environment.
This device is designed to measure blood pressure and pulse rate of people for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremes in temperature, humidity, direct
sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened with water and a neutral detergent. Neveruse alcohol, benzene, thinner or other harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff for long periods, as such treatment may shorten the life of the components.
The device and cuff are not water resistant. Prevent rain, sweat and water from soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions, microwave ovens, cellular telephones,
X-ray or other devices with strong electrical fields.
English 2

Used equipment, parts and batteries are not treated as ordinary household waste, and must be disposed ofaccording to the applicable local regulations.
When reusing the device, confirm that the device is clean.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the wrist must be squeezed by the cuff hard enough to temporarily stop blood flow
through the artery. This may cause pain, numbness or a temporary red mark to the wrist. This condition willappear especially when measurement is repeated successively.
Any pain, numbness, or red marks will disappear with time.

Contraindications
The following are precautions for proper use of the device.
Do not apply the cuff to an arm with another medical electrical equipment attached. The equipment may not
function properly.
People who have a severe circulatory deficit in the arm must consult a doctor before using the device, to avoid
medical problems.
Do not self-diagnose the measurement results and start treatment by yourself. Always consult your doctor forevaluation of the results and treatment.
Do not apply the cuff on an arm with an unhealed wound.
Do not apply the cuff on an arm receiving an intravenous drip or blood transfusion. It may cause injury oraccidents.
Do not use the device where flammable gases such as anesthetic gases are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as a high-pressure oxygen chamber oran oxygen tent. It may cause a fire or explosion.
English 3Parts Identification
Display
Systolic pressure
Diastolic pressure
Pulse rateBattery cover
Carrying case
START button

Alkaline batteries(LR03 or AAA)

Cuff

Part of display
MEMORY
Pressure bar indicator
andWHO classification indicator
I.H.B indicator(Irregular heartbeat indicator)
Battery indicator

English 4

Systolic pressure
Diastolic pressure
Pulse rate
Heart markSymbols
Symbols

Function / Meaning
Recommended Action
Standby and Turn the device on
Battery installation guide
Direct current
SN
Serial number
Date of manufacture
Type BF: Device, cuff and tubing are designed to provide
special protection against electrical shocks.
The indicator that appears while measurement is inMeasurement is in progress. Remain as still asprogress. It blinks when the pulse is detected.
possible.
Irregular Heartbeat indicator. (I.H.B.)
The indicator that appears when an irregular heartbeat or any
excessive body movement is detected during the measurement.
Previous measurements stored in MEMORY.
Full Battery The battery power indicator during measurement.
Replace all batteries with new ones when theLow Battery The battery power is low when it blinks.
indicator blinks.
Unstable blood pressure due to movement during theTry measurement again.
measurement.
Remain very still during the measurement.
The systolic and diastolic values are within 10 mmHg ofeach other.
Fasten the cuff correctly, and try measurement
The pressure value did not increase during inflation.
again.
The cuff is not fastened correctly.
EThe pulse is not detected correctly.
English 5

Symbols
SYS
DIA
PUL./min

Function / Meaning
Systolic blood pressure in mmHg
Diastolic blood pressure in mmHg
Pulse per minuteEC directive medical device label

Recommended Action

WEEE label
Manufacturer
EU-representative
Refer to instruction manual/booklet

English 6

Using the Monitor
1. Installing / Changing the Batteries
1. Slide the battery cover as shown to remove it from the device.
2. Insert two AAA / LR03 size batteries into the battery compartment asshown, taking care that the polarities (+ and -) are correct.
3. Slide the battery cover back on the device.
• Insert the batteries as shown in the battery compartment or on the cuff. If installed incorrectly, the device will not work.
• When(LOW BATTERY mark) appears on the display, replace both batteries with new ones at the sametime. Do not mix old and new batteries. It may shorten the battery life, or cause the device to malfunction.
•(LOW BATTERY mark) does not appear when the batteries are drained.

• The battery life varies with ambient temperature and may be shorter at low temperatures.
• Remove the batteries if the device is not to be used for a long period of time. The batteries may leak and cause amalfunction.
• Use the specified batteries only. The batteries provided with the device are for testing the device performanceand may have a limited life.

2. Fastening the Cuff
1. Wrap the cuff around your wrist about 1 cm above your hand as shown in the figure atthe right.
2. Fasten the cuff tightly using the Velcro strip.
Note: For proper measurements, fasten the cuff tightly and measure on a bare wrist.
About 1 cm

English 7

Using the Monitor
3. How to Take Accurate Measurements
For the most accurate blood pressure measurement:• Sit comfortably on a chair. Rest your arm on the table. Do not cross your legs. Keep yourfeet on the floor and straighten your back.
• Relax for about 5 to 10 minutes before measurement.
• Raise your hand so that the cuff is at the same level as your heart.
• Remain still and keep quiet during measurement.
• Do not measure immediately after physical exercise or a bath.
Rest for twenty or thirty minutes before taking measurement.
• Try to measure your blood pressure at the same time every day.

4. Measurement (Operation)
During measurement, it is normal for the cuff to feel very tight.
Select a measurement mode (operation of the device).
Normal measurement with storing data : ............Press the START 1 or START 2 button.
Unrecorded measurement : .................................Press both START buttons simultaneously.
Recalling data : .....................................................Press and hold the START 1 or START 2 button to recall data.
Clearing data : .......................................................Remove the batteries and re-install them.

5. After Measurement
After the measurement has completed, remove the cuff and record your data, then press the START 1 or START 2button to turn off the power.
Note: The device has an automatic power shut-off function, which turns the power off approximately one minute aftermeasurement.
Allow at least three minutes between measurements on the same person.
English 8Measurements
Note: Model UB-512 can store the last thirty sets of data for each button. We recommend that if two people regularly use thedevice, one uses the START 1 button and the other uses START 2 button, to prevent confusion over stored data.

Normal Measurement with Storing Data
1. Sit comfortably with the cuff at the same level as your heart andrelax.
2. Press the START 1 (or START 2 ) button.
The average recorded blood pressures and data number are
displayed briefly.
Then the display changes as indicated in the figure on the rightas measurement begins.
The cuff starts to inflate. It is normal for the cuff to feel very tight.
Note: If you wish to stop inflation at any time, press either START
button again.
3. The measurement starts automatically when pressurizing starts,and the(heart mark) blinks.
4. When measurement is complete, the systolic and diastolicpressure readings and pulse rate are displayed.
The cuff exhausts the remaining air and completely deflates
automatically.
5. Press either START button again to turn off the power.
Remove the cuff.
Note: The device is provided with an automatic power shut-offfunction.
Allow at least three minutes between measurements on thesame person.
English 9START

buttonAveraged
dataData
numberPressurizingandMeasurementin progress

Systolicpressure
Diastolicpressure
WHO
Classification
Pulse rate
I.H.B indicator

Exhausts remaining
air automaticallyMeasurements
Notes for Accurate Measurement
Sit down in a comfortable position. Place the arm to be used for the measurement on a table or other support sothat the center of the cuff will be at the same height as your heart.
Relax for about five to ten minutes before taking a measurement. If you are excited or depressed by emotional
stress, the measurement will reflect this stress as a higher (or lower) than normal blood pressure reading and thepulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you are doing and what you have eaten.
What you drink can have a very strong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very weak or irregular heartbeat, the device
may have difficulty determining your blood pressure.
Should the device detect a condition that is abnormal, it will stop the measurement and display an error symbol.
See page 5 for the description of symbols.
This blood pressure monitor is intended for use by adults only. Consult with your physician before using thisdevice on a child. A child should not use this device unattended.

Operation Mode
1. Normal measurement with storing data (Refer to page 9)
Press the START 1 (or START 2 ) button. Blood pressure ismeasured and data is stored within the device. The device canstore the last thirty sets of data for each button. We recommendthat if two people regularly use the device, one uses the START 1button and the other uses START 2 button.
2. Recalling data
Press and hold the START 1 (or START 2 ) button to recall the data.
Stored data are automatically displayed in order from the last data.
English 10

Data
numberStoreddata

Press and hold theSTART button

Alternation

3. Clearing data
Remove batteries to clear all data. Re-install them again.
4. Unrecorded measurement (Guest mode)
Press both of the START 1 and START 2 buttons at once.
An unrecorded measurement is started.

What is an Irregular Heartbeat?
Model UB-512 blood pressure monitor provides a blood pressure and pulse rate measurement even when anirregular heartbeat occurs. An irregular heartbeat is defined as a heartbeat that varies by 25% from the averageof all heartbeats during the blood pressure measurement. It is important that you be relaxed, remain still and do
not talk during measurements.
Note: We recommend contacting your physician if you see thisindicator frequently.

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing

Inflation in progress

Measurement in progress
English 11Inflation completeWHO Classification Indicator
Each six segments of the bar indicator correspond to the WHO blood pressure classification described on the next page.
ExampleModeratehypertension

WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimalMildhypertension

High normal: The indicator displays a segment, based on thecurrent data, corresponding to the WHO classification.

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of the arteries. Systolic pressure occurs when theheart contracts. Diastolic pressure occurs when the heart expands. Blood pressure is measured in millimeters ofmercury (mmHg). One's natural blood pressure is represented by the fundamental pressure, which is measured
first thing in the morning while one is still at rest and before eating.

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left unattended, can cause many health problems including
stroke and heart attack. Hypertension can be controlled by altering lifestyle, avoiding stress, and with medication undera doctor’s supervision.
English 12To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?

WHO Blood Pressure Classification

Standards to assess high blood pressure, without regard to age,have been established by the World Health Organization (WHO),as shown in the chart on the right.

Diastolic blood pressure

Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 Reference Material: Journal of Hypertension 1999, Vol 17 No.2
mmHg higher than that measured at home. Home measurement
Grade 3 hypertension (severe)
reduces the effects of outside influences on blood pressure
Grade 2 hypertension (moderate)
readings, supplements the doctor's readings and provides a moreaccurate, complete blood pressure history.
Grade 1 hypertension (mid)High-normal
NormalOptimalSystolic blood pressure

An individual’s blood pressure varies greatly on a daily and seasonal basis. It mayvary by 30 to 50 mmHg due to various conditions during the day. In hypertensive
individuals, variations are even more pronounced. Normally, the blood pressure rises
while at work or play and falls to its lowest levels during sleep. So, do not be overly
concerned by the results of one measurement.
Take measurements at the same time every day using the procedure describedin this manual to get to know your normal blood pressure. Regular readings givea more comprehensive blood pressure history. Be sure to note date and time
when recording your blood pressure. Consult your doctor to interpret your bloodpressure data.
English 13

Blood Pressure

Blood Pressure Variations
Typical fluctuation within a day(Measured every five minutes)
Systolic

DiastolicSleepTroubleshooting
ProblemPossible Reason
Batteries are drained.

Nothing appears in thedisplay, even when the Battery terminals are not in the correct
power is turned on.
position.
The cuff does notinflate.

The device does notmeasure.
Readings are too highor too low.

Other

Battery voltage is too low.
(LOW BATTERY mark) blinks. If thebatteries are drained completely, the markdoes not appear.
The cuff is not fastened properly.
You moved your arm or body during themeasurement.

Recommended Action
Replace all batteries with new ones.
Reinstall the batteries with negative andpositive terminals matching those indicated on
the battery compartment.
Replace both batteries with new ones.

The cuff position is not correct.

Fasten the cuff correctly.
Make sure you remain very still and quiet
during the measurement.
Sit comfortably and still.
Raise your hand so that the cuff is at the samelevel as your heart.
If you have a very weak or irregular heat beat,the device may have difficulty in determining
your blood pressure.

The value is different from that measured at aclinic or doctor’s office.

See “Why measure blood pressure at home?”.

Remove the batteries. Place them backproperly and try measurement again.
Note: If the actions described above do not solve the problem, contact the dealer. Do not attempt to open or repair
this product yourself, as any attempt to do so will make your warranty invalid.

English 14

MaintenanceDo not open the device. It uses delicate electrical components and an intricate air unit that could be damaged. If you
cannot fix the problem using the troubleshooting instructions, request service from your dealer or from the A&D
service group. The A&D service group will provide technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life. However it is generally recommended to havethe device inspected every 2 years, to ensure proper functioning and accuracy. Please contact either yourauthorized dealer or A&D for maintenance.

Technical DataType

UB-512

Measurement Oscillometric measurement
methodMeasurement Pressure: 20 - 280 mmHg
range
Pulse:
40 - 180 beats / minute
Measurement Pressure: ±3 mmHg
accuracy
Pulse:±5%
Power supply 2 x 1.5V alkaline batteries (LR03 or AAA)Approx. 400 measurements, when AAA alkaline
Number ofbatteries are used, with pressure value ofmeasurements 170 mmHg at room temperature of 23°C

Applied part Cuff Type BF
Useful life
Device: 5 years (when used six times a day)
Clinical test According to ANSI / AAMI SP-10 1992
EMC

IEC 60601-1-2: 2007

MemoryOperatingconditions
Transport/storage
conditions

Last 30 measurements for each button+10°C to +40°C / 15%RH to 85 %RH
800hPa to 1060hPa-10°C to +60°C / 15%RH to 85 %RH

WristDimensions Approx. 63 [W] x 56 [H] x 21 [D] mmcircumference 13.5 - 21.5 cm
Internally powered ME equipment
Classification Continuous operation mode
WeightApprox. 82 g, excluding the batteries
Note: Specifications are subject to change for improvement without prior notice.
EMC table information is listed on our website: http://www.aandd.jp/products/manual/medical/emc_en.pdf
English 151-243 Asahi, Kitamoto-shi, Saitama 364-8585 JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
A&D INSTRUMENTS LIMITED

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United KingdomTelephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.

1756 Automation Parkway, San Jose, California 95131 U.S.A.
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119
A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409