Ambulatory Blood Pressure Monitor
TM-244x series
All-in-One

TM-2441

TM-2440

The TM-2440 and TM-2441 have been validated in accordance with ISO protocols. Lightweight andcompact design facilitates easy operation by patients and research participants. New features allow for
the detection of Irregular Heartbeat (IHB) as well as standard ABPM measurements. The TM-2441 also
has an LCD screen and slide switch to enable easy self-measurement. Collects data on various
external factors (temperature, pressure, etc.) that influence blood pressure fluctuations. Included
software provides useful analysis from data collected.

The global and regional guidelines recommend Out-of-Office measurements
TM-2441

TM-2440

External Dimensions

W × H × D (cm)

66 × 24.5 × 95

66 × 24.5 × 95

WeightWithout batteries

135 g

120 g

Power Supply

Alkaline or Ni-MH "AA" batteries

2 × LR6("AA")2 × LR6("AA")Memory Data

Number of datasets

600

600

BP Measurement
Function

ABPM
Self-Measurement

444–––

IHB

Irregular Heart Beat

44

Record Pulse wave

Use analysis software

44Multi sensor

Activity / Temperature / Atmosphericpressure

4–––

Data communication

USB1.1 compliant
Bluetooth (SDK available for MedicalDeveloper)

4

4

4–––

Display

LCD: Self -BP measurement display
OLED: ABPM display

44–––
4

Clinical trial

ISO 81060-2:2013

44

Ambulatory
Blood Pressure
MonitorTM-2441

All-in-One

0123

TM-2441 can process five types of self-measurement while
recording environmental data (temperature, pressure,activity) through a multi-sensor. It is an “All-in-One” modelthat covers any environment from the doctor’s office to home.
Pulse data during BP measurement is recorded in the device,with analysis software (included) providing the precisewaveform. Viewing this data improves the accuracy of BP
measurements and helps to verify irregular pulse waves 1.
n 24-hour ambulatory blood pressure monitoringn Self-measurement- OBP: Office Blood Pressure- AOBP : Automated Office Blood Pressure- HBP: Home Blood Pressure- ANBP : Automated Night Blood Pressure▼Self measurement mode displays
LCD display: This large display shows data frommeasurements in self-measurement mode.
You can easily check the time until the next
ABPM measurement in addition to values inself-measurement mode. There are also icons
for battery level, Bluetooth status, IHB detection,clock, mode and memory status.

- ASBP : Automated Self Blood Pressuren Easy to use mode slide switch
n The pulse correction algorithm for faster measurement &noise rejectionn Battery Power (2 alkaline or Ni-MH "AA")n Output to data to analysis software (included with device)n USB & BLE 2 Connectivityn Ingress protection IP22n Protocol ISO 81060-2
1) Hypertension Canada
2) SDK available for Medical Developer

TM-2440

A&D’s TM-2440 is an entry-level model designed for ABPM
measurement. Its compact body simplifies the monitoring ofresearch participants and comes standard with an IrregularHeartbeat (IHB) function, which detects pulse irregularities
and can check waveforms easily via analysis software(included). These features in the TM-2440 support ABPM
measurement.
n 24-hour ambulatory blood pressure monitoringn Light weight & compact size
120 g*, 66(W)×24.5(H)×95(D) mm *Without batteriesn OLED display (ABPM mode)n The pulse correction algorithm for faster measurement &noise rejectionn Battery powered (2 alkaline or Ni-MH “AA”)
n Output of data to analysis software (included with device)n USB connectivityn Ingress protection IP22n Protocol ISO 81060-20123

Software ABPM Data Manager Included with every unit .............................
▼Multisensor data analysis with the software

The newly designed software can analyze environmental
data collected by the multi-sensor in addition to ABPM
measurement data. Temperature, pressure and activity canbe graphed alongside blood pressure. TM-2441 only(TM-2440 does not have a multi-sensor)▼Waveform data recording and displaying

The TM-244x series records blood pressure as anoscillometric waveform. With our new analysis software you
can check the waveform record for each measurement.
Quickly determine the accuracy of measured values.

▼System requirements
1 GHz or higher processor
Computer CPU:
2 GB or more
Memory :Operation System(recommended) : Windows 10 64-bit version or higher
Recommended 800×600 pixels or more
SVGA :
DVD drive (drive corresponding to enclosed media)
Disk :
Available space of 24 GB
Hard disk :Options ................................................................................................
Cuff

Cuff cover

Cuff cloth

Carrying holder

Clip

TM-CF502A
TM-CF402A
TM-CF302A
TM-CF202A
TM-CF802A
TM-CF306A

Extra large cuff for left arm (36-50cm)
Large cuff for left arm (28-38cm)
Adult cuff for left arm (20-31cm)
Small cuff for left arm (15-22cm)
Adult cuff for right arm (20-31cm)
Disposable cuff (10 sheets)

AX-133024503-S
AX-133024663-S
AX-133024500-S
AX-133024667-S
AX-133024353-S

Extra large cuff cover for left arm (10 sheets)
Large cuff cover for left arm (10 sheets)
Adult cuff cover for left arm (10 sheets)
Small cuff cover for left arm (10 sheets)
Adult cuff cover for right arm (10 sheets)

AX-133025103-S
AX-133025102-S
AX-133024487-S
AX-133025101-S
AX-133025104-S

Extra large cloth for left arm (2 sheets)
Large cuff cloth for left arm (2 sheets)
Adult cuff cloth for left arm (2 sheets)
Small cuff cloth for left arm (2 sheets)
Adult cuff cloth for right arm (2 sheets)

AX-133025995
AX-110B-20-S

Carrying holder
Clips (5 pieces)Ambulatory Blood Pressure Monitor

Specifications

TM-2441/TM-2440

TM-2441TM-2440

Measurement Method

Oscillometric measurement method

Pressure detection method

Semiconductor pressure sensor

Pressure display range

0 to 299 mmHg (299 mmHg or more is hidden )Measurement accuracy

Pressure : ±3 mmHg
Pulse rate : ±5 %Minimum display division

Pressure : 1 mmHg
Pulse rate : 1 beat / minute

Measurement range

Systolic pressure : 60 to 280 mmHg
Diastolic pressure : 30 to 160 mmHg
Pulse rate :30 to 200 beat / minutePressurization method

Micro pump

Automatic pressurization

85 to 299 mmHg

Interval Setting
DisplayIntervals at each section which divides 24 hours up to maximum six parts.
Interval : OFF, 5, 10, 15, 20, 30, 60, 120 minutes
A-BPM : OLED, 96 × 39 pixels, white characters
S-BPM : LCD, 40 × 50 mm, Display

Clock

A-BPM : OLED, 96 × 39 pixels, white characters

24 hour clock

Measurement count

200 times or more. varies due to measurement conditions.

Memory Data

600 data max

Power supply

With the same type of batteries : 2 × 1.5V batteries (LR6 or AA size)
Alkaline battery or Nickel-hydrogen battery (Ni-MH) 1900 mAh or more
Backup battery for built-in clock : Lithium rechargeable coin cell battery ML2016

Rated voltage
InterfaceDC 2.4 V and DC 3.0 V
USB : USB1.1 compliant.
Cable length : 1.5 m or shorter

USB : USB1.1 compliant.

Micro-USB B type terminal can connect todedicated peripheral (using standard driver software).
Bluetooth Ver.4.1 (BLE) :Cable length : 1.5 m or shorter
Micro-USB B type terminal can connect todedicated peripheral (using standard driver software).

Wireless device can be connected.
SDK available for Medical Developer
Operating conditions

Temperature : +10 to +40 °CHumidity :30 to 85 %RH (no condensation)

Atmospheric pressure bothfor operation and storage
condition

700 to 1060 hPa

External Dimensions
WeightApprox. 95 (L) × 66 (W) × 24.5 (H) mm
Approx. 135 g (excluding batteries)

Approx. 120 g (excluding batteries)

A&D Company Limited.(JAPAN)

A&D AUSTRALASIA PTY LTD(Australia)

A&D ENGINEERING, INC.(USA)A&D RUS CO., LTD.(Russia)

A&D Instruments Canada Inc.(CANADA)

A&D Technology Trading(Shanghai)Co., Ltd(China)

A&D INSTRUMENTS LIMITED(United Kingdom)

A&D INSTRUMENTS INDIA PRIVATE LIMITED(India)

URL:aandd.jp

URL:andonline.com

URL:autocontrol.com

URL:andprecision.com

URL:andaustralasia.com.au
URL:and-rus.ru

URL:aanddtech.cn

URL:aanddindia.in

TM2441-ADCC-03-PR03-245001GP* REF*ADC-TM2441