Waterproof Static Weighing Indictor

FG-D-CWP

Affordable Stainless Steel Indicator Ideal for
Building Various Types of Waterproof Scales
A new solution for scale makers is here. This highly washable, alcohol cleanable,
stainless steel indicator helps in not just designing scales that are easy to keep
hygienic but also helps in making weighing highly accurate and productive inmany different ways.
 IP67 dust & waterproof, stainless steel (SUS304) construction

In accordance with the IP67 rating of the IEC/EN standard 60529, the FG-D-CWP prevents theintrusion of water even while immersed to a depth of 1 meter for 30 minutes. It also permits noparticulate matters, which can damage the device, inside the housing.
The stainless steel surface is resistant to rust/corrosion and scratches, and even allows cleaning
with alcohol (diluted to 70 % or less) as well as rinse with water.

 Drive capacity of up to four 350 Ω load cells with a remote sensing function

The FG-D-CWP is suitable for building various sizes of scales from a small bench scale that uses asingle-point load cell to a large floor scale that uses four load cells.
The remote sensing function minimizes error by compensating for the effect of resistance inwires, which is especially effective when using an extended length of load cell cable orconnecting multiple load cells via a summing box.
(Continued on next page)Other Useful Features
 Freely configurable capacity and scale division (max. display resolution: 1/100,000)
 Clearly-visible, large LCD digits (30 mm character height) with white backlight Bright and multi-colored LED comparator lights for intuitive HI/OK/LO judgments
 Auto-tare and negative comparison functions to realize "take-away" check weighing Power supply from either the provided AC adapter, USB port on another device, or off-the-shelfmobile battery via USB cable (3 m)
 Capable of both data communication and power supply at the same time when connected to a PC
 Multiple units of measure: kg, g, lb, oz, t and pcs (for counting mode) Stand for setting the indicator on the floor or up on a wall at an arbitrary angle
 Password lock and key restriction functions to avoid
unwanted changes to settings
 Adjustable response characteristics, Display holdfunction, Auto power ON/OFF, GLP/GMP output, etc.

Multi-colored LED comparator lights

Specifications
Input sensitivity

0.1 μV/d min. (d: scale division)

Input signal range-16 mV to 16 mV

Load cell excitation

DC 5 V ±10 %, Remote sensing function included
Up to 4 load cells (350 Ω) can be connected

Temperature coefficients

Zero: ±0.02 μV/˚C typ., ±0.1 μV/˚C max.
Span: ±3 ppm/˚C typ., ±15 ppm/˚C max.

Nonlinearity±0.01 % of full scale

Display resolution

100,000 d max.

Display

7 segment LCD with white backlight (character height: 30 mm)

Display refresh rate

Approx. 10 times/second

Operating environment-10 ˚C to 40 ˚C / 14 ˚F to 104 ˚F, 85 % R.H. or less (no condensation)

Power supply

AC adapter*i, USB port*ii or mobile battery via USB Type-A
USB cable length: Approx. 3 m

Dimensions

240 (W) × 76 (D) × 155 (H) mm

Net weight

Approx. 1.6 kg (including the stand)*i Provided as standard*ii Data communication is also possible when connected to a USB port of a PC.

Discover Precision
A&D Company, Ltd. (JAPAN)

A&D Korea Ltd. (South Korea)

URL: aandd.jp

URL: andk.co.kr

A&D Engineering, Inc. (USA)A&D Rus Co., Ltd. (Russia)

URL: andonline.com

URL: and-rus.ru

A&D Australasia Pty Ltd. (Australia)

A&D Instruments India (P) Ltd. (India)

URL: andaustralasia.com.au

URL: aanddindia.in

A&D Instruments Ltd. (United Kingdom)

A&D Scientech Taiwan Ltd. (Taiwan)

URL: andprecision.com

URL: aandd.com.tw<German Sales Office>A&D Instruments Thailand Ltd. (Thailand)

URL: andprecision.com

URL: thai.andprecision.com