Software IRBIS® 3

Thermal Stress Analysis with
Lock-in Thermography

ImageIR® 8300

Universal Thermography Camera for Continuous Operation Applications
Detector Format
Large detector enables
highest sensivity

640×512205 HzIR-Frame Rate
Analysis of extreme temperaturechanges and gradients in full frame
Measurement Accuracy
Highly accurate andrepeatable measurements
Thermal Resolution
Precise detection of smallest
temperature differences
Rotating Filter Wheel and Aperture Wheel
Enables measurement tasks with high object
temperatures and spectral thermography
Motor Focus
Precise, fast and remotely controllable;
including multiple autofocus functions

With its ImageIR® 8300, InfraTec introduces another thermographic camera model belonging to the ImageIR® high-end
camera series. The implementation of a (640 × 512) IR pixel
MWIR detector allows 205 Hz full-frame real-time imaging without compromising any thermal accuracy. The ImageIR® 8300
and its cooled focal-plane array photon detector reach an outstanding thermal resolution better than 0.02 K. The new version
was developed for most demanding operations in ­research anddevelopment and process monitoring fields. Its modular structure consisting of the optical, detector and interface section,
makes the camera easily compatible to the related applications
and for tailored configurations. An inte­grated trigger interface
guarantees a repeatable high-precision triggering of quick procedures. Multiple configurable digital inputs and outputs serveas control ports for the camera or as generator of digital control
signals for external devices.
The optical channel consists of the exchangeable infrared lensas well as application-specific apertures, filters and reference
elements. All exchangeable ImageIR® 8300 standard lenses
can be combined with a motorised focus unit easily operablefrom the camera‘s application software. It allows precise, fastand remotely controlled motorised focusing and is part of theautofocus function.

Process and Trigger Interface
Highly precise repeatable data recording;time- and event-controlled
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range

MCT: (1.5 … 5.5) µm
InSb: (1.5 … 5.7) µm

Pitch

15 µm

Detector

MCT or InSb

Detector format (IR pixels)(640 × 512)

Image acquisition

SnapshotReadout mode

ITR / IWR

Aperture ratio

f/3.0 or f/2.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 1,500) °C, up to 3,000 °C*

Measurement accuracy± 1 °C or ± 1%

Temperature resolution @ 30 °C

MCT: Better than 0.02 K
InSb: Better than 0.025 K

Frame rate (full / half / quarter / sub frame)*

MCT: Up to 151 / 540 / 1,520 / 2,807 Hz
InSb: Up to 205 / 570 / 1,020 / 5,000 Hz

Window mode

Yes

Focus

Manual, motorised or automatic*

Dynamic range

Up to 16 bit*

Integration time

(0.6 … 20,000) µs

Rotating filter wheel*

Up to 6 positionsRotating aperture wheel*

Up to 5 positions

Interfaces

GigE, CAMLink*, HDMI*

Trigger

4 IN / 2 OUT, TTL

Analogue signals*, IRIG-B*

2 IN / 2 OUT, yes

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5

Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight

MCT: (241 × 120 × 160) mm*; InSb: (235 × 120 × 160) mm*
3.3 kg (without lens)

Further functions

Multi Integration Time*

Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on model

Lenses

Focal length (mm)

FOV (°)

Wide-angle lens12

(43.6 × 35.5)

1.3Standard lens25

(21.7 × 17.5)

0.6

Telephoto lens

50

(11.0 × 8.8)

0.3

Telephoto lens

100

(5.5 × 4.4)

0.15

Telephoto lens

200

(2.7 × 2.2)

0.08

Macro and microscopic lensesIFOV (mrad)

Minimum object distance (mm)Object size (mm)Pixel size (µm)

Close-up for telephoto lens 50 mm

300(58 × 46)

90

Close-up for telephoto lens 100 mm

500(48 × 38)

75

Microscopic lens M=1.0×40 / 195 / 300(9.6 × 7.7)

15

Microscopic lens M=3.0×22(3.2 × 2.6)

5

Microscopic lens M=8.0×14(1.2 × 0.96)

1.9© InfraTec 04 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.