Format-filling image of circuit board
and detailed zoom into image

ImageIR® 10300

Full HD Thermography Camera – Highest Geometric Resolution
Detector Format
Efficient measurement of smallest
details on large-scale objects
960
×768

400 HzIR-Frame Rate
Analysis of extreme temperaturechanges and gradients
Measurement Accuracy
Highly accuracte andrepeatable measurements
Thermal Resolution
Precise detection of smallest
temperature differences
10 GigE Interface
High-speed, long-distance interference
proof data transmission
High-speed Mode
Increase frame rates and thermal resolution
at the same time using binning technology

With its detector format of (1,920 × 1,536) IR pixels the ImageIR® 10300 sets new standards in geometric resolution worldwide and creates thermograms with an unprecedented image
detail and sharpness. For the first time an infrared camera
permits Full HD images for civil use with a cooled focal-planearray photon detector. In combination with the small pitch dimension of 10 µm, this ensures that measurement, inspection
and surveillance tasks can be solved even more efficiently thanbefore. For example, wherever such very fine structures need tobe analysed on large-surface measurement objects, users savetime, effort and thus costs.
An integrated trigger interface guarantees a repeatable highprecision triggering of quick procedures. Despite the detector
format of about 3 Megapixels, the transfer of full frame images
achieves a rate up to 113 Hz. In subframe format even considerably higher frame rates are realizable. Thanks to the 10 GigE interface of the ImageIR® 10300, users can store large amounts ofmeasurement data on a computer in the shortest amount of time.
The interface is a part of the modular design of the entire highend camera series ImageIR®. Individual adjustments can easily
be realised like retrofitting a remotely controllable filter and aperture wheel or a motor focus unit. A broad variety of infraredlenses with highest optical performance parameters provides thecamera’s outstanding thermal sensitivity.

Complete Optical Assortment
Adaptation of the image geometry
to almost every measuring situation
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range(3.6 … 4.9) µm

Pitch

10 µm

DetectorInSb

Detector format (IR pixels)(1,920 × 1,536)

Image acquisition

SnapshotReadout mode

ITR / IWR

Aperture ratio

f/2.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 1,200) °C, up to 3,000 °C*

Measurement accuracy± 1 °C or ± 1 %Temperature resolution @ 30 °C

Better than 0.035 K / 0.022 K in high-speed mode

Frame rate (full  / half  / quarter  / sub frame)*

Up to 113 / 216  / 396  / 1,915 Hz;
High-speed mode: up to 400 / 692  / 1,088  / 2,493 Hz

Window mode

YesFocus

Manual, motorised or automatic*

Dynamic range

13 bitIntegration time

(1 … 20,000) µs

Rotating aperture wheel and filter wheel*

Up to 6 positions

Interfaces

10 GigE, HDMI*

Trigger

4 IN / 2 OUT, TTL

Analogue signals*, IRIG-B*

2 IN / 2 OUT, yes

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5

Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight(241 × 123 × 160) mm; 4.7 kg (without lens)

Further functions

High-speed mode*, Multi Integration Time*

Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on model

Lenses

Focal length (mm)

FOV (°)

Wide-angle lens25

(42.0 × 34.2)

0.4Standard lens

50

(21.7 × 17.5)

0.2

Telephoto lens

100

(11.0 × 8.8)

0.1

Macro and microscopic lenses

IFOV (mrad)

Minimum object distance (mm)Object size (mm)Pixel size (µm)

Close-up for telephoto lens 50 mm

300(115 × 92)

60

Close-up for telephoto lens 100 mm

500(96 × 77)

50

Microscopic lens M=1.0×40(19 × 15)

10

Microscopic lens M=8.0×14(2.4 × 1.92)

1.3© InfraTec 04 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.