Precision Health Scale

Instruction Manual
Mode d'emploi

1WMPD4002776A

Manuale d’uso

Italiano

Bedienungsanleitung

DeutschManual instructivo

Español Français English

UC-351PBT-Ci

Contents
Startup Guide ............................................................................................. 1
Safety Precautions..................................................................................... 2
Wireless Communication Function.......................................................... 3
Accessories................................................................................................ 3
Parts Names ............................................................................................... 4
Installing (or Replacing) the Batteries ..................................................... 5
Registering with a Receiver (Pairing) ...................................................... 6
Skip Registering with a Receiver (Pairing).............................................. 7
Setting the Unit .......................................................................................... 8
Measuring Your Weight ............................................................................. 9
Other Functions ....................................................................................... 10
Specifications........................................................................................... 12
Before Requesting Service ..................................................................... 13
Maintaining the Scale .............................................................................. 13

Dear Customers
Congratulations on purchasing a state-of-the-art A&D health scale, one of the most advanced healthscales available today. Designed for ease of use and accuracy, this device will facilitate your daily
weight management regimen. We recommend that you read through this manual carefully before
using the device for the first time.

1WMPD4002776A

Startup Guide
Get Started

Install the batteries (P.5)

Pair with a receiver (P.6)
Measure without pairing with a receiver (P.7)

Set the Unit (P.8)

Measuring(P.9)

Step on the scaleConfirm that StEP and oFF are alternately displayed

Step off the scaleConfirm your weightkg※ Data is transmitted.
Data is stored in the scale if there is no receiver. (P.10)● Receivers can also be unpaired. (P.10)

English 1

Safety Precautions
The following items contain information to prevent the risk of harm or injury to yourself or other people.
Please read these items carefully and be sure to use this product correctly.

DangerThis symbol indicates "an immediate danger which could result
in death or serious injury".

Warning

This symbol indicates "the possibility of death or serious injury".

CautionThis symbol indicates "the possibility of physical injury or
damage to property".

Precautions for UseGeneral Precautions

Warning
Do not rely on self-diagnosis when attempting weight loss or therapeutic exercise.
Follow the instructions of a physician or specialist.
Do not step on the edges of the scale or jump on the scale.
It may result in a fall, causing injury. The impact may also cause damage to the scale.
Do not use the scale on an uneven surface.
It may damage the scale. Also, the scale cannot measure accurately.
Do not use the scale on a slippery surface, or step on the scale with wet feet.
It may result in slipping and injury. Also, the scale cannot measure accurately.

CautionDo not place the scale where it is exposed to excess water or humidity such as a bathroom, where it is exposed todirect sunlight, where it is exposed directly to the air outlet of an air conditioning unit, or near fire.
It may cause the scale to malfunction.
This scale should only be used for weighing body weight.
This scale is a precision measuring device. Do not drop the scale, or expose it to vibration or powerful shocks.
Do not disassemble, repair or modify the scale.
It may cause malfunction or injury.
Persons with disabilities should be assisted by a caregiver when using this scale.
Failure to do so may result in a fall, causing injury.
When installing the batteries, be sure that the + (positive) and - (negative) terminals are correct.
Installing incorrectly may cause overheating, battery fluid leaks or rupturing of the batteries, resulting in damage tothe scale or injury.
This scale is designed for personal use.
Store this scale out of the reach of small children.
Do not store this scale vertically. If for any reason this scale is stored vertically, be sure not to let the scale fall.
Do not place any items on this scale during storage.
If use of this scale is shared with a person who has an infectious disease such as a skin disease on the feet, etc.,
clean the scale before use.
(For details on cleaning the scale, see "Maintaining the Scale (P.13)".)
To accurately measure your weight, use the scale on a flat, hard floor. Do not use on a carpet or other soft surface.

English 2

Wireless Communication Function
This scale uses the Bluetooth wireless to communicate with a receiver device. At the end of
measurement, the measurement data is automatically sent from this scale to the receiver device.
Bluetooth is communication technology that enables devices to communicate wirelessly. For details on
Bluetooth, see "Scale Communication (P.11)".

Receiver Device
This is a device such as personal computer or smartphone on which an application to enable
Bluetooth communication with this scale has been downloaded and installed.

Precautions for Wireless Function
Warning
Do not use in places where wireless communication is prohibited, such as on airplanes or in hospitals.
This scale may have an adverse effect on electronic devices or medical electrical equipment.

CautionIf the communication function of this scale is used near a wireless device that communicates in the 2.4 GHz range,
the processing speed of both devices may decrease. Do not use near a microwave oven where a magnetic field,
static electricity or radio wave interference may occur. (The signal may not reach due to environmental reasons.)※ A&D cannot accept any responsibility for any losses incurred such as operating malfunctions or lossof data that may occur through the use of this scale.
※ This scale is not guaranteed to connect to all Bluetooth compatible devices.

Accessories
If any of the accessories are missing or damaged, contact the product vendor or an A&D Sales Branch.
Precision Health Scale

Instruction Manual
Mode d'emploi

Manuale d’uso

Italiano

Bedienungsanleitung

DeutschManual instructivo

Español Français English

UC-351PBT-Ci

1WMPD4002776

Instruction Manual

English 3

Alkaline Batteries x 4

Parts Names
Scale
Battery compartment
HandleSetting Switch

Feet

Display Section
BluetoothCommunication
SymbolStable Symbol
Battery indicator

Unit

Display Symbols

Display Section
Symbol○Name

Description
This symbol is displayed when the measurement
display is stable.
The normal battery power indicator with charge
remaining.

Ready/Stable symbol
Battery indicator↓
Low battery indicator

If this symbol is displayed, replace all four batteries
with new ones.

Bluetooth Communication Symbol
Scale Underside
SymbolName

Description

WEEE label-----

English 4

Installing (or Replacing) theBatteries
the battery compartment cover on
1. Openthe underside of the scale.

four new AA size alkaline batteries
2. Installwith the positive (+) and negative (-)
terminals properly aligned with thecorresponding symbols in the battery
compartment.
the battery compartment cover.
3. Close
The scale switches to pairing mode when batteries are
installed.
To pair the scale with a receiver, see "Registering with a
Receiver (Pairing) (P.6)". If you do not want to pair the scalewith a receiver, see "Skip Registering with a Receiver(Pairing) (P.7)".

CautionInstall each battery with the positive (+) and negative (-) terminals properly aligned with thecorresponding symbols in the battery compartment.
Improper installation of the batteries may cause battery leakage.
Replace all four batteries with new ones, when thesymbol is displayed.
Do not mix new and used batteries. It may cause damage to the scale or shorten the battery life.
Be sure to use AA size alkaline batteries. Do not use any other type of battery.
Never use rechargeable batteries with this scale.
The battery life is dependant upon the environmental temperature, battery type and usage.
The batteries provided with the device are for testing the scale performance and may have alimited life.
Remove the batteries from the battery compartment, when the scale is not to be used for a longtime. Storing the scale with the batteries installed may result in battery fluid leaking from thebatteries, causing damage to the scale. The supplied batteries are for checking operation of thescale and may have a short lifespan.

Automatic Power-Off Function
The scale has an "Automatic power-off" function.
After the measurement results are displayed for 15 seconds, the scale is automatically turned off.
English 5

Registering with a Receiver(Pairing)
This scale needs to be registered with a receiver in order to use Bluetooth communication with thereceiver. The process of registering this scale with a receiver is known as pairing.
When pairing is completed, measurement data from this scale is automatically sent to the paired device
each time a measurement is made when the receiver is in the state in which it can receive transmissions.
Consult the receiver instruction manual and set it to enable pairing. If the receiver has a "pairing wizard",
use the wizard to complete pairing.
Place the receiver and this scale as close as possible to carry out pairing.

four AA size alkaline batteries into
1. Install
this scale.
For details, see "Installing (or Replacing) the Batteries (P.5)".

2. Confirm that "Pr" is displayed.

≪♪Beep≫" " flashes.
This scale is discoverable by the receiver.

Pairing can be skipped using the following method.
i） Press the setting switch when "Pr" is displayed.

3. Start pairing on the receiver.

Follow the pairing instructions for the receiver to search for,select and pair with this scale.
If a PIN code is requested by the receiver, enter "39121440".

4. Confirm that "End" is displayed.

≪♪Beep beep≫

Pairing completed.
If "E-10" or "E-11" is displayed, perform the pairing procedureagain from step 1.

※ This scale can be paired with two receivers.
※ If this scale is paired with a third receiver, the information of the receiver that was communicated withmost recently remains.
※ Once the scale paired with the receiver, it is no need to perform pairing after replacing batteries.

Cautions for Pairing
Turn the power of other Bluetooth devices off when performing pairing.
Multiple devices cannot be paired at the same time.
If measurement data cannot be received, try pairing again.
After pairing is complete, all communications are done automatically.
To unpair a receiver, see "Unpairing (P.10)".
In the unlikely event that this scale causes radio wave interference to another wireless station,either change the location where the scale is used or stop using the scale immediately.

English 6

Skip Registering with a Receiver(Pairing)
four AA size alkaline batteries into
1. Install
this scale.
For details, see "Installing (or Replacing) the Batteries (P.5)".

the setting switch when "Pr" is
2. Press
displayed.
Press the setting switch to turn the power off.

English 7≪♪Beep≫

Setting the Unit
that the scale is
1. Confirm
turned off, and press
and hold the setting
switch.

≪♪Beep≫When the setting switch iscontinuously pressed, "unit" and"uPr" are alternately displayed.

the setting
2. Releaseswitch when "unit" is≪♪Beep≫displayed.
The display changes from "unit" todisplay the unit that is currently set.
kg

the setting switch
3. Pressto select the unit thatkg

you want to set.
The unit changes each time thesetting switch is pressed.
Wait for 3 seconds after setting the
unit.

lb

that "End" is
4. Confirm
displayed.

≪♪Beep beep≫

The settings are saved.
After three seconds of
non-operation, the scale confirms
the selected unit with two beeps,
displays “End ” and automaticallyturns off.

English 8

Measuring Your Weight
Step On Weighingthat the scale is
1. Confirm
turned off, and step on
the scale gently.
Measurement starts when you stepon the scale.
A progress bar is displayed during≪♪Beep≫measurement. Stand up straight andremain as still as possible during
measurement.

off the scale when
2. Step"StEP" and "oFF" are≪♪Beep beep beep≫alternately displayedwith three beeps.
If you do not step off, the scale
beeps every two seconds.
The scale turns off automaticallyafter 2 minutes.
All display segments are indicatedwhen you step off the scale.
Do not touch the scale while all
display segments are indicated.
It may cause misreading.

in
c
lb
kg

the
3. Confirm
measurement results.

kg

The scale turns off automatically 15
seconds after the measurement
results are displayed.
Data is automatically sent to thereceiver if the receiver is in thereceivable state.
If the scale is not paired with areceiver, the data is stored in thescale memory.
(See "Transmitting Temporarily
Stored Data (P.10)".)Data stored in the scale memory
cannot be viewed on this scale.
You can also shut the power off bypressing the setting switch.

English 9

Other Functions
Unpairingthat the scale is
1. Confirm
turned off, and press andhold the setting switch.
When the setting switch iscontinuously pressed, "unit" and"uPr" are alternately displayed.

≪♪Beep≫the setting
2. Releaseswitch when "uPr" is≪♪Beep≫displayed.

that "End" is
3. Confirm
displayed.

≪♪Beep beep≫

The receiver is unpaired.
If "E-10" or "E-11" is displayed,
perform the unpairing procedureagain from step 1.
The scale turns off automatically.

Transmitting Temporarily Stored Data
If measurement data cannot be sent to a receiver, this scale can temporarily store data of a maximum of
200 measurements in the built-in memory. When the scale is able to transmit to a receiver, all of the datais transmitted. If the amount of data temporarily stored exceeds 200 measurements, the oldest data isdeleted and replaced with new data.
※ Stored data cannot be viewed on this scale.
※ The maximum amount of data that can be temporarily stored may differ depending on the receiver.
English 10

Scale Communication
This scale is equipped with a Continua certified Bluetooth wireless communication function.
Measurement data can be transmitted by connecting to a Bluetooth device that is compatible with
Continua connection specifications.
Devices that are Continua certified carry the Continua logo mark.
Bluetooth devices carry the Bluetooth logo mark.
For details on pairing this scale with a receiver that is equipped with Bluetooth, see "Registering witha Receiver (Pairing) (P.6)".
Also refer to the pairing instruction manual of the receiver that you want to pair with.

Ensuring a Good Connection
Ensure the space between this scale and the receiver is unobstructed. The communication distance
may be shortened due to the building construction or other obstructions. In particular, communication
may not be possible through reinforced concrete walls, etc.
Do not use the Bluetooth connection near a wireless LAN or other wireless equipment, near devices
that emit electro-magnetic waves such as microwave ovens, locations that have a large number ofobstructions, or other poor environments for radio waves. It may cause the connection to be
frequently interrupted, communication speeds to drop substantially, and errors to occur.
When used near an IEEE802.11g/b/n wireless LAN device, the devices cause mutual radio wave
interference, and the communication speed may drop or connection may be impossible. In thissituation, turn the power of the device that you are not using off, or change the location where it isused.
If the scale does not communicate properly when used near a wireless device or broadcast station,change the location where it is used.

CautionThe communication distance between this scale and the receiver depends on the Bluetooth output
class of the receiver.
For Class1 Bluetooth receivers: Max. 100 m
For Class2 Bluetooth receivers: Max. 10 m
This distance may become shorter due to the surrounding environment; therefore check that the
distance is close enough for data to be transmitted after the measurement.

Time Settings
This scale has a built-in clock. The date and time of measurement are included in the measurement data.
The clock is synced with the clock in the receiver. For details on the clock specifications, see theinstruction manual of the receiver.
English 11

Specifications
Model

UC-351PBT-Ci

Maximum capacity

150 kg / 330 lb

Minimum display

0.1 kg units from 0 to 100.0 kg
0.2 kg units from 100.0 to 150.0 kg
0.2 lb units from 0 to 220.0 lb
0.5 lb units from 220.0 to 330.0 lb

Wireless
communication

WML-C40AH (MITSUMI ELECTRIC CO., LTD.)
Bluetooth Ver.2.1 class 1
SSP HDP Compatible
Frequency band: 2402 MHz to 2480 MHz
Maximum RF output power: 20 dBm

Setting Items

Units: kg or lb

Power source

Four AA size alkaline batteries

Battery life

Approx. 1 year (two measurements per day when alkaline batteries are used)

Dimensions

350 (W) ×350(D) ×39 (H) mm

WeightApprox. 2.3 kg (including batteries)

Accessories

Instruction manual, Alkaline batteriesCompliance with European Directives
The scale complies with the statutory EMC (Electromagnetic Compatibility)
Directive 2014/30/EU.
Hereby, MITSUMI ELECTRIC CO., LTD. declares that the radio equipment type
WML-C40AH is in compliance with Directive 2014/53/EU. The full text of the EU declarationof conformity is available at the following internet address:
http://www.aandd.jp/products/manual/manual_medical.html
Notes: Indoor use only. The displayed value may be adversely affected under extreme
electromagnetic influences.


Compliance with FCC Rules
The scale complies with part 15 of the FCC rules and contains the FCC ID POOWML-C40.

Compliance with Industry Canada
IC: 4250A-WMLC40.

English 12

Before Requesting Service
Symptoms

Corrective Action

E is displayed

Exceeds the maximum capacity (150 kg / 330 lb).

E-0 is displayed

Exceeds the maximum capacity (150 kg / 330 lb).

Err is displayed

Exceeds the maximum capacity (150 kg / 330 lb).

E-10 is displayed

Pairing or unpairing has not completed correctly. Remove and reinstall
the batteries.

E-11 is displayed
H is displayed
is displayed
Nothing is displayed when
the setting switch ispressed

Exceeds the maximum capacity (150 kg / 330 lb).
Remove and reinstall the batteries. Or replace all four batteries with new
AA size alkaline batteries.

The measurement
results are greatly
different from theexpected values.

Place the scale on a solid level surface.

Measurement doesn't startwhen I step on the scale.

Did you press the setting switch when stepping on the scale?
Place the scale on a hard floor, and press the setting switch before
measuring.

Remove any foreign objects that might be touching the bottom of thescale.

Maintaining the ScaleIf the scale becomes dirty, soak a cloth in water or mild detergent, wring the cloth out thoroughly andthen wipe the scale clean.
After wiping away the dirt with the damp cloth, dry the scale using a soft, dry cloth.
Never use alcohol, benzene, thinner or other harsh chemicals to clean the scale.
Remove the batteries from the battery compartment, when the scale is not going to be used for a longtime

English 13

Table des matières
Guide de démarrage .................................................................................. 1
Consignes de sécurité .............................................................................. 2
Fonction de communication sans fil ....................................................... 3
Accessoires................................................................................................ 3
Nomenclature ............................................................................................. 4
Mise en place (ou remplacement) des piles............................................ 5
Enregistrement sur un récepteur (Pairage)............................................. 6
Ignorer l'enregistrement sur un récepteur (Pairage).............................. 7
Réglage de l'appareil ................................................................................. 8
Mesure de votre poids............................................................................... 9
Autres fonctions ...................................................................................... 10
Spécifications........................................................................................... 12
Avant de requérir une maintenance....................................................... 13
Entretien du pèse-personne ................................................................... 13

Chers clients
Nous vous félicitons d'avoir acheté un pèse-personne médical de pointe A&D, l'un despèse-personnes médicaux les plus évolués actuellement disponibles. Conçu pour la facilitéd'utilisation et la précision, cet appareil simplifiera la gestion de votre poids au quotidien. Nous vous
recommandons de lire soigneusement ce manuel avant d'utiliser l'appareil pour la première fois.

1WMPD4002776A

Guide de démarrage
Prise en mains

Mettre les piles en place (P.5)

Procéder au pairage avec un récepteur (P.6)
Procéder à la mesure sans pairage avec un récepteur (P.7)

Réglage de l'appareil (P.8)

Mesure(P.9)

Monter sur le pèse-personne

Vérifier que StEP et oFF s'affichent en alternance

Descendre du pèse-personne

Confirmez votre poidskg※ Les données sont transmises.
Les données sont stockées dans le pèse-personne s'il n'existe pas de récepteur
externe. (P.10)● Le pairage des récepteurs peut également être annulé. (P.10)

Français 1

Consignes de sécurité
Les éléments suivants comportent des informations afin d'éviter tout risque de dommage de blessurespour vous-même ou d'autres personnes. Veuillez lire soigneusement ces éléments et vous assurerd'utiliser correctement ce produit.

DangerCe symbole indique « un danger immédiat qui pourrait
provoquer la mort ou de graves blessures ».
Ce symbole indique « un risque de mort ou de graves

Avertissement blessures ».
Attention

Ce symbole indique « un risque de blessure physique ou dedommage au bien ».

Précautions d'utilisation
Précautions générales

Avertissement
Ne vous fiez pas à un autodiagnostic si vous essayez de perdre du poids ou de faire des exercices thérapeutiques.
Conformez-vous aux instructions d'un médecin ou d'un spécialiste.
Ne montez pas sur les bords du pèse-personne et ne sautez pas dessus.
Vous pourriez chuter et vous blesser. L'impact pourrait également endommager le pèse-personne.
N'utilisez pas le pèse-personne sur une surface inégale.
Cela pourrait l'endommager. En outre, le pèse-personne ne pourrait pas procéder correctement à la mesure.
N'utilisez pas le pèse-personne sur une surface glissante, et ne montez pas dessus avec les pieds mouillés.
Vous pourriez glisser et vous blesser. En outre, le pèse-personne ne pourrait pas procéder correctement à lamesure.

Attention
Ne placez pas le pèse-personne à un endroit où il est exposé à un excès d'eau ou d'humidité, tel qu'une salle debains, à la lumière directe du soleil, à une sortie d'air de climatiseur ou à proximité du feu.
Cela pourrait provoquer un dysfonctionnement.
Ce pèse-personne ne doit être utilisé que pour la mesure de poids corporel.
Ce pèse-personne est un appareil de mesure de précision. Ne laissez pas tomber le pèse-personne, ne l'exposezpas à des vibrations ou à des chocs intenses.
Vous ne devez pas démonter, réparer ni modifier le pèse-personne.
Cela pourrait provoquer un dysfonctionnement ou des blessures.
Les personnes handicapées doivent être aidées par une personne soignante.
Sinon, elles pourraient tomber et se blesser.
Lors de la mise en place des piles, assurez-vous de respecter la polarité des bornes + (positives) et - (négatives).
Une mise en place incorrecte pourrait provoquer une surchauffe, une fuite du liquide des piles ou la rupture decelles-ci et entraîner des dommages au pèse-personne ou provoquer des blessures.
Ce pèse-personne est conçu pour une utilisation personnelle.
Rangez-le hors de portée des enfants en bas âge.
Ne le rangez pas verticalement. Si pour une raison quelconque ce pèse-personne est rangé verticalement,
assurez-vous qu'il ne tombera pas.
Ne placez aucun objet sur ce pèse-personne lorsqu'il est rangé.
Si vous partagez l'utilisation de ce pèse-personne avec une personne ayant une maladie infectieuse, telle qu'une
maladie de peau au niveau des pieds, etc., nettoyez-le avant usage.
(Pour de plus amples informations sur le nettoyage, reportez-vous à la section « Entretien du pèse-personne(P.13) ».)Pour mesurer précisément votre poids, utilisez le pèse-personne sur un sol plat et dur. Ne l'utilisez pas sur de lamoquette ou toute autre surface molle.

Français 2

Fonction de communication sans fil
Ce pèse-personne utilise le système sans fil Bluetooth pour communiquer avec un récepteur. À la fin d'une
mesure, les données de la mesure sont automatiquement envoyées de ce pèse-personne au récepteur.
Bluetooth est une technologie de communication qui permet à des périphériques de communiquer sansfil. Pour de plus amples informations sur Bluetooth, reportez-vous à la section « Communication dupèse-personne (P.11) ».

Récepteur
Il s'agit d'un périphérique tel qu'un ordinateur personnel ou un smartphone sur lequel une application
permettant d'activer la communication Bluetooth avec ce pèse-personne a été téléchargée et installée.

Précautions relatives à la fonction sans fil
Avertissement
Ne l'utilisez pas dans des endroits où les communications sans fil sont interdites tels que dans des avions ou dansdes hôpitaux.
Ce pèse-personne peut avoir des effets négatifs sur des appareils électroniques ou des équipements médicauxélectriques.

Attention
Si la fonction de communication de ce pèse-personne est utilisée à proximité d'un appareil sans fil qui communique
sur la bande de 2,4 GHz, il est possible que la vitesse de traitement des deux appareils diminue. Ne l'utilisez pas à
proximité d'un four à micro-ondes où des interférences dues à un champ magnétique, à de l'électricité statique ou àdes ondes radio pourraient se produire. (Le signal pourrait ne pas atteindre son but pour des raisonsenvironnementales.)※ A&D n'assume aucune responsabilité quant à des pertes découlant de dysfonctionnements liés àl'utilisation ou à des pertes de données qui pourraient survenir suite à l'utilisation de ce
pèse-personne.
※ La connexion de ce pèse-personne à tous les périphériques compatibles Bluetooth n'est pas garantie.

Accessoires
Si l'un des accessoires manque ou est endommagé, contactez le fournisseur du produit ou la succursalede vente d'A&D.
Precision Health Scale

Instruction Manual
Mode d'emploi

Manuale d’uso

Italiano

Bedienungsanleitung

DeutschManual instructivo

Español Français English

UC-351PBT-Ci

1WMPD4002776

Mode d'emploi

Piles alcalines x 4

Français 3

NomenclaturePèse-personne
Logement des piles
Poignée
Commutateur de réglage

Pieds

Section de l'affichage
Symbole decommunication
BluetoothSymbole stable
Témoin des piles

Unité

Symboles de l'affichage
Section de l'affichage
Symbole○Nom
Symbole Prêt/Stable
Témoin des piles↓
Témoin de piles faibles

Clignotant

Description
Ce symbole s'affiche lorsque l'affichage de lamesure est stable.
Témoin normal de puissance des piles avec charge
restante.
Si ce symbole s'affiche, remplacez les quatre piles
par des neuves.

Symbole de communication BluetoothFace inférieure du pèse-personne
SymboleNom

Description

Étiquette WEEE-----

Français 4

Mise en place (ou remplacement) des piles
le couvercle du logement des1. Ouvrez
piles situé sur la face inférieure dupèse-personne.

en place quatre piles alcalines
2. Mettez
neuves au format AA en respectant lespolarités positives (+) et négatives (-)
indiquées par les symboles correspondants
dans le logement des piles.

le couvercle du logement des piles.
3. RefermezLe pèse-personne passe en mode de pairage lorsque despiles sont mises en place.
Pour procéder au pairage du pèse-personne avec unrécepteur, reportez-vous à la section « Enregistrement sur unrécepteur (pairage) (P.6) ». Si vous ne souhaitez pas
procéder au pairage du pèse-personne avec un récepteur,reportez-vous à la section « Ignorer l'enregistrement sur unrécepteur (pairage) (P.7) ».

Attention
Mettez en place chacune des piles en respectant les polarités positives (+) et négatives (-)
indiquées par les symboles correspondants dans le logement des piles.
Une mise en place inappropriée des piles pourrait provoquer une fuite de celles-ci.
s'affiche.
Remplacez les quatre piles par des neuves lorsque le symbole
Ne mélangez pas des piles neuves et des piles usagées. Cela pourrait endommager lepèse-personne ou réduire la durée de vie des piles.
Veillez à utiliser des piles alcalines au format AA. N'utilisez aucun autre type de piles.
N'utilisez pas de piles rechargeables avec ce pèse-personne.
La durée de vie des piles dépend de la température de l'environnement, du type de piles et del'utilisation.
Les piles fournies avec l'appareil sont destinées à tester les performances du pèse-personne etpeuvent avoir une durée de vie limitée.
Retirez les piles du logement des piles lorsque le pèse-personne n'est pas utilisé pendant une
période prolongée. Le stockage du pèse-personne avec les piles en place pourrait provoquer une
fuite du liquide des piles et endommager le pèse-personne. Les piles fournies sont destinées à lavérification du fonctionnement du pèse-personne et peuvent avoir une durée de vie réduite.

Fonction d'extinction automatique
Le pèse-personne dispose d'une fonction d'« extinction automatique ».
Une fois le résultat de la mesure affiché pendant 15 secondes, le pèse-personne s'éteint
automatiquement.
Français 5

Enregistrement sur un récepteur(Pairage)
Ce pèse-personne doit être enregistré sur un récepteur pour utiliser la fonction de communication
Bluetooth avec le récepteur. Le processus d'enregistrement de ce pèse-personne sur un récepteur est
appelé pairage.
Une fois le pairage terminé, les données de mesure de ce pèse-personne sont automatiquement
envoyées au périphérique appairé chaque fois qu'une mesure est réalisée lorsque le récepteur se trouveen état de recevoir les transmissions.
Consultez le mode d'emploi du récepteur et configurez-le pour activer le pairage. Si le récepteur disposed'un « assistant de pairage », utilisez-le pour effectuer le pairage.
Placez le récepteur et ce pèse-personne aussi près que possible l'un de l'autre pour procéder aupairage.

en place quatre piles alcalines au
1. Mettez
format AA dans ce pèse-personne.
Pour de plus amples informations, reportez-vous à la section« Mise en place (ou remplacement) des piles (P.5) ».

si « Pr » s'affiche.
2. Vérifiez« » clignote.

≪♪Bip≫

Ce pèse-personne peut être trouvé par le récepteur.
Il est possible d'ignorer le pairage grâce à la méthodesuivante.
i) Appuyez sur le commutateur de réglage lorsque « Pr »s'affiche.

le pairage sur le récepteur.
3. Démarrez
Suivez les instructions de pairage pour le récepteur à rechercher,sélectionnez ce pèse-personne et procédez à son pairage.
Si un code PIN est requis par le récepteur, saisissez« 39121440 ».

4. Vérifiez si « End » s'affiche.

≪ ♪Bip bip ≫Le pairage est terminé.
Si « E-10 » ou « E-11 » s'affiche, répétez la procédure depairage depuis l'étape 1※ Il est possible de procéder au pairage de ce pèse-personne avec deux récepteurs.
※ Si ce pèse-personne est appairé à un troisième récepteur, les informations du récepteur avec lequelune communication a été établie pour la dernière fois demeurent.
※ Une fois le pèse-personne appairé au récepteur, il n'est plus nécessaire de procéder au pairage
après le remplacement des piles.

Français 6

Précautions relatives au pairage
Lorsque vous procédez au pairage, coupez l'alimentation des autres périphériques Bluetooth.
Il est impossible de procéder au pairage simultané de plusieurs périphériques.
S'il est impossible de recevoir les données de mesure, procédez à nouveau au pairage.
Une fois le pairage terminé, toutes les communications s'effectuent automatiquement.
Pour annuler le pairage d'un récepteur, reportez-vous à « Annulation du pairage (P.10) ».
Dans l'éventualité peu probable où ce pèse-personne provoquerait des interférences au niveau
des ondes radio avec une autre station sans fil, changez l'emplacement où ce pèse personne est
utilisé ou cessez immédiatement de l'utiliser.

Ignorer l'enregistrement sur unrécepteur (Pairage)en place quatre piles alcalines au
1. Mettez
format AA dans ce pèse-personne.
Pour de plus amples informations, reportez-vous à la section« Mise en place (ou remplacement) des piles (P.5) ».

sur le commutateur de réglage
2. Appuyez
lorsque « Pr » s'affiche.
※ Appuyez sur le commutateur de réglage pour couper
l'alimentation.

Français 7≪♪Bip≫

Réglage de l'appareil
que le1. Vérifiez
pèse-personne est
allumé, et appuyez demanière prolongée sur lecommutateur de réglage.

≪♪Bip≫

Si vous appuyez en continu sur lecommutateur de réglage, « unit »et « uPr »s'affichent en alternance.

le commutateur
2. Relâchez
de réglage lorsque≪♪Bip≫« unit » s'affiche.
L'affichage passe de « unit » àl'unité actuellement configurée.
kg

sur le3. Appuyez
commutateur de réglagekg

pour sélectionner l'unité
que vous souhaitez
configurer.

lbL'unité change à chaque fois quevous appuyez sur le commutateurde réglage.
Patientez 3 secondes après avoir
configuré l'unité.

si « End »4. Vérifiez
s'affiche.

≪♪Bip bip≫Les réglages sont enregistrés.
Après trois secondes sansutilisation, le pèse-personne
confirme l'unité sélectionnée àl'aide de deux bips, affiche « End »et s'éteint automatiquement.

Français 8

Mesure de votre poids
Mesure lorsque vous montez dessus
que le1. Vérifiez
pèse-personne est allumé etmontez doucement dessus.

La mesure démarre lorsque vousmontez sur le pèse-personne.
Une barre de progression s'affiche≪♪Bip≫

pendant la mesure. Tenez-vous droit
et demeurez aussi immobile que
possible pendant la mesure.

du pèse-personne
2. Descendez
lorsque « StEP » et « oFF »≪♪Bip bip bip≫s'affichent alternativement en
faisant entendre trois bips.

Si vous ne descendez pas, lepèse-personne émet des bips
toutes les deux secondes.
Le pèse-personne s'éteint
automatiquement après 2 minutes.
Tous les segments de l'affichagesont affichés lorsque vous
descendez du pèse-personne.
Ne touchez pas au pèse-personne
lorsque tous les segments sontaffichés.
Cela pourrait provoquer une
mauvaise mesure.

in
c
lb
kg

les résultats
3. Confirmez
de la mesure.

kg

Le pèse-personne s'éteint
automatiquement 15 secondes après
l'affichage des résultats de la mesure.
Les données sont automatiquement
envoyées au récepteur si celui-ci setrouve en état de réception.
Si le pairage du pèse-personne avec
un récepteur n'est pas effectué, lesdonnées sont stockées dans lamémoire du pèse-personne.
(Reportez-vous à « Transmission desdonnées temporairement stockées(P.10) ».)Les données stockées dans lamémoire du pèse-personne ne sont
pas consultables sur celui-ci.
Vous pouvez également couper
l'alimentation en appuyant sur lecommutateur de réglage.

Français 9

Autres fonctions
Annulation du pairageque le1. Vérifiez
pèse-personne estéteint, et appuyez demanière prolongée sur lecommutateur de réglage.

≪ ♪Bip ≫Si vous appuyez en continu sur lecommutateur de réglage, « unit » et
« uPr »s'affichent en alternance.

2. Relâchez le commutateur≪ ♪Bip ≫3. Vérifiez si « End » s'affiche.

≪ ♪Bip bip ≫de réglage lorsque« uPr » s'affiche.

Le pairage du récepteur est annulé.
Si « E-10 » ou « E-11 » s'affiche,
répétez la procédure d'annulation
du pairage depuis l'étape 1.
Le pèse-personne s'éteint
automatiquement.

Transmission des données temporairement stockées
S'il est impossible d'envoyer les données de mesure à un récepteur, ce pèse-personne peut
temporairement stocker des données correspondant à un maximum de 200 mesures dans la mémoireintégrée. Lorsque le pèse-personne est en mesure d'établir une transmission avec un récepteur, toutes
les données sont transmises. Si la quantité de données temporairement stockées dépasse 200 mesures,les données les plus anciennes sont effacées et remplacées par les nouvelles données.
※ Il est impossible de consulter les données stockées sur ce pèse-personne.
※ La quantité maximale de données qui peuvent être stockées temporairement peut différer en fonctiondu récepteur.
Français 10

Communication avec le pèse-personne
Ce pèse-personne est équipé d'une fonction de communication sans fil Bluetooth homologuée Continua.
Les données des mesures peuvent être transmises en connectant un périphérique Bluetooth compatible
avec les spécifications de connexion Continua.
Les périphériques homologués Continua portent le logo Continua.
Les périphériques Bluetooth morte le logo Bluetooth.
Pour de plus amples informations sur le pairage de ce pèse-personne avec un récepteur compatible
Bluetooth, reportez-vous à « Enregistrement sur un récepteur (pairage) (P.6) ».
Reportez-vous également au mode d'emploi du pairage du récepteur que vous souhaitez appairer.

Pour s'assurer d'une bonne connexion
Assurez-vous que l'espace entre ce pèse-personne et le récepteur est dégagé. La distance decommunication peut être réduite à cause de la construction du bâtiment ou d'autres obstructions. En
particulier, la communication peut s'avérer impossible à travers des parois en béton armé, etc.
N'utilisez pas la connexion Bluetooth à proximité d'un réseau local sans fil ou d'un autre équipementsans fil, à proximité de périphériques qui émettent des ondes électromagnétiques tels que des fours à
micro-ondes, dans des lieux comprenant un grand nombre d'obstructions ou d'autres environnements
peu appropriés aux ondes radio. Cela pourrait provoquer de fréquentes interruptions de la connexion,
une réduction importante des vitesses de communication et des erreurs.
En cas d'utilisation à proximité d'un périphérique de réseau local sans fil IEEE802.11g/b/n, lesappareils provoquent des interférences radio mutuelles et la vitesse de communication peut chuter ou
la communication peut s'avérer impossible. Dans ce cas, éteignez l'appareil que vous n'utilisez pas
ou changez l'endroit dans lequel vous l'utilisez.
Si le pèse-personne ne communique pas correctement lorsqu'il est utilisé à proximité d'unpériphérique sans fil ou d'une station de diffusion, changez l'endroit dans lequel vous l'utilisez.

Attention
La distance de communication entre ce pèse-personne et le récepteur dépend de la classedémission Bluetooth du récepteur.
Pour les récepteurs Bluetooth de classe 1 : Max. 100 m
Pour les récepteurs Bluetooth de classe 2 : Max. 10 m
Cette distance peut diminuer à cause de l'environnement ; par conséquent, vérifiez que la distanceest suffisamment faible pour que les données puissent être transmises après la mesure.

Réglage de l'heure
Ce pèse-personne est doté d'une horloge intégrée. La date et l'heure des mesures sont incluses dansles données des mesures. L'horloge est synchronisée avec l'horloge du récepteur. Pour de plus amples
informations sur les spécifications de l'horloge, reportez-vous le mode d'emploi du récepteur.

Français 11

Spécifications
ModèleUC-351PBT-Ci

Capacité maximale

150 kg / 330 lb

Affichage minimal

Unités de 0,1 kg de 0 à 100,0 kg
Unités de 0,2 kg de 100,0 à 150,0 kg
Unités de 0,2 lb de 0 à 220,0 lb
Unités de 0,5 lb de 220,0 à 330,0 lb

Communicationsans filWML-C40AH (MITSUMI ELECTRIC CO., LTD.)
Bluetooth Ver.2.1 classe 1
Compatible SSP HDP
Bande de fréquence: 2402 MHz à 2480 MHz
Puissance de sortie RF maximale: 20 dBm

Éléments de réglage

Unités : kg ou lb

Source d'alimentation

Quatre piles alcalines au format AA

Durée de vie des piles Environ 1 an (à raison de deux mesures par jour lorsque des piles alcalines
sont utilisées)Dimensions

350 (L) ×350 (P) ×39 (H) mm

Poids

Environ 2,3 kg (piles incluses)

Accessoires

Mode d'emploi, piles alcalines

Conformité aux directives européennes
Le pèse-personne est conforme à la directive obligatoire CEM (Compatibilité
Électromagnétique) 2014/30/UE.
Le soussigné, MITSUMI ELECTRIC CO., LTD., déclare que l'équipement radioélectrique dutype WML-C40AH est conforme à la directive 2014/53/UE. Le texte complet de la déclaration
UE de conformité est disponible à l'adresse internet suivante:
http://www.aandd.jp/products/manual/manual_medical.html
Remarques : Utilisation en intérieur exclusivement. La valeur affichée peut être altérée dans desconditions d'influences électromagnétiques extrêmes.


Conformité aux réglementations de la FCC
Le pèse-personne est conforme à la partie 15 des règles de la FCC et possède l'ID FCC POOWML-C40.


Conformité à Industrie Canada
IC : 4250A-WMLC40.

Français 12

Avant de requérir une maintenanceSymptômes

Signification / Action corrective

E s'affiche

Surcharge : dépassement de la capacité maximale de pesage de labalance (150 kg / 330 lb).

E-0 s'affiche

Surcharge : dépassement de la capacité maximale de pesage de labalance (150 kg / 330 lb).

Err s'affiche

Surcharge : dépassement de la capacité maximale de pesage de labalance (150 kg / 330 lb).

E-10 s'affiche

Le pairage ou l'annulation du pairage ne s'est pas terminé correctement.
Retirez et remettez en place les piles.

E-11 s'affiche
H s'affiches'affiche
Rien ne s'affiche lorsque
vous appuyez sur lecommutateur de réglage
Les résultats de la mesuresont très différents desvaleurs prévues.

Dépasse la capacité maximale (150 kg / 330 lb).
Retirez et remettez en place les piles. Ou remplacez les quatre piles par
des piles alcalines neuves au format AA.

Placez le pèse-personne sur une surface solide et de niveau.
Retirez tous les corps étrangers qui pourraient toucher la partie inférieuredu pèse-personne.

La mesure ne démarre pas Avez-vous appuyé sur le commutateur de réglage lorsque vous êteslorsque vous montez sur le monté sur le pèse-personne ?
pèse-personne.
Placez le pèse-personne sur un sol dur et appuyez sur le commutateurde réglage avant de procéder à la mesure.

Entretien du pèse-personne
Si le pèse-personne est sale, trempez un chiffon dans un peu d'eau ou un détergent léger, essorez à
fond le chiffon, puis essuyez le pèse-personne pour le nettoyer.
Après avoir nettoyé la saleté à l'aide du chiffon humide, séchez le pèse-personne à l'aide d'un chiffon
doux et sec.
N'utilisez jamais d'alcool, de benzène, de diluant ni d'autres produits chimiques durs pour nettoyer lepèse-personne.
Retirez les piles du logement des piles lorsque le pèse-personne n'est pas utilisé pendant une
période prolongée.

Français 13

Contenido
Guía de inicio ............................................................................................. 1
Precauciones de seguridad ...................................................................... 2
Función de comunicación inalámbrica ................................................... 3
Accesorios.................................................................................................. 3
Nombres de las partes .............................................................................. 4
Instalación (o sustitución) de las pilas .................................................... 5
Registro con un receptor (conexion)....................................................... 6
Omisión del registro con un receptor (conexion) .................................. 7
Ajuste de la unidad .................................................................................... 8
Medición del peso...................................................................................... 9
Otras funciones........................................................................................ 10
Especificaciones...................................................................................... 12
Antes de solicitar asistencia técnica ..................................................... 13
Mantenimiento de la báscula .................................................................. 13

Estimado cliente
Le felicitamos por la compra de una moderna báscula de precisión A&D, una de las básculas
disponibles más avanzadas en la actualidad. Este dispositivo, preciso y sencillo de usar, facilitará su
régimen de control de peso diario. Le recomendamos leer detenidamente este manual antes de usar
el dispositivo por primera vez.

1WMPD4002776A

Guía de inicio
Procedimientos

Instale las pilas (P.5)

Conecte la báscula con un receptor (P.6)
Realice la medición sin conectarse con un receptor (P.7)

Ajuste de la unidad (P.8)

Medición(P.9)

Súbase a la báscula

Confirme que StEP y oFF se visualizan alternativamente

Bájese de la báscula

Confirme su pesokg

※ Los datos se transmiten.
Si no hay ningún receptor, los datos se almacenan en la báscula. (P.10)● Los receptores también pueden desconectarse. (P.10)

Español 1

Precauciones de seguridad
Los siguientes elementos contienen información destinada a evitar el riesgo de que usted u otras
personas sufran daños o lesiones. Léalos detenidamente y asegúrese de utilizar este producto
correctamente.

PeligroEste símbolo indica "un peligro inmediato que podría
provocar la muerte o lesiones graves".

Advertencia

Este símbolo indica "la posibilidad de producirse la muerte olesiones graves".

Precaución

Este símbolo indica "la posibilidad de producirse lesiones
físicas o daños materiales".

Precauciones de usoPrecauciones generales

Advertencia
No confíe en el autodiagnóstico cuando intente perder peso o hacer ejercicio terapéutico.
Siga las instrucciones de un médico o especialista.
No se sitúe sobre los bordes de la báscula ni salte encima de ella.
Si lo hace, podría caerse y sufrir lesiones. El golpe también podría producir daños en la báscula.
No utilice la báscula sobre una superficie irregular.
Podrían producirse daños en la báscula. Además, la báscula no proporcionaría mediciones precisas.
No utilice la báscula sobre una superficie resbaladiza, ni se suba a ella con los pies mojados.
Si lo hace, podría resbalar y sufrir lesiones. Además, la báscula no proporcionaría mediciones precisas.

Precaución
No coloque la báscula en lugares donde quede expuesta a un exceso de agua o de humedad (por ejemplo, unbaño), en lugares donde quede expuesta a la luz directa del sol o directamente expuesta a la salida de aire de unaparato de aire acondicionado, o cerca del fuego.
Si lo hace, podría producirse un mal funcionamiento de la báscula.
Esta báscula debe usarse únicamente para medir el peso corporal.
Esta báscula es un dispositivo de medición de precisión. No deje caer la báscula ni la exponga a vibraciones ogolpes fuertes.
No desmonte, repare ni modifique la báscula.
Si lo hace, podrían producirse lesiones o un mal funcionamiento de la báscula.
Las personas con discapacidad deben usar esta báscula con la ayuda de un cuidador.
De lo contrario, podrían caerse y sufrir lesiones.
Cuando instale las pilas, compruebe que los polos + (positivo) y - (negativo) estén colocados correctamente.
Una instalación incorrecta puede producir sobrecalentamiento, fuga de fluido de las pilas o rotura de las pilas,
provocando daños en la báscula o lesiones.
Esta báscula se ha diseñado para uso personal.
Almacene esta báscula en un lugar fuera del alcance de los niños pequeños.
No almacene esta báscula en posición vertical. Si por algún motivo esta báscula se almacena en posición vertical,
asegúrese de que no se caiga.
No coloque ningún objeto sobre esta báscula mientras esté almacenada.
Si esta báscula se usa de forma compartida con una persona que tenga una enfermedad infecciosa, como por
ejemplo una enfermedad cutánea en los pies, limpie la báscula antes de usarla.
(Para obtener más información sobre cómo limpiar la báscula, consulte "Mantenimiento de la báscula" (P.13).)
Para medir el peso con precisión, utilice la báscula sobre un suelo duro y plano. No la utilice sobre una alfombra omoqueta ni ninguna otra superficie blanda.

Español 2

Función de comunicación
inalámbrica
Esta báscula utiliza una conexión inalámbrica Bluetooth para comunicarse con un dispositivo receptor.
Al finalizar la medición, los datos de dicha medición se envían automáticamente desde la báscula aldispositivo receptor.
Bluetooth es una tecnología de comunicación que permite a los dispositivos comunicarse de forma inalámbrica.
Para obtener más información sobre la tecnología Bluetooth, consulte "Comunicación de la báscula" (P.11).

Dispositivo receptor
El dispositivo receptor es un dispositivo (por ejemplo, un ordenador personal o un smartphone) en elque se ha descargado e instalado una aplicación que permite establecer la comunicación Bluetoothcon esta báscula.

Precauciones de uso de la función inalámbrica
Advertencia
No utilice esta función en lugares donde la comunicación inalámbrica esté prohibida, como por ejemplo en aviones o
en hospitales.
Esta báscula puede tener un efecto negativo sobre el funcionamiento de dispositivos electrónicos o equipos
eléctricos médicos.

PrecauciónSi la función de comunicación de esta báscula se utiliza cerca de un dispositivo inalámbrico que se comunique en la
gama de 2,4 GHz, ello podría disminuir la velocidad de procesamiento de ambos dispositivos. No utilice esta función
cerca de un horno microondas ni en lugares donde pueda haber un campo magnético, electricidad estática ointerferencias de ondas de radio . (La señal podría no llegar correctamente por motivos ambientales.)※ A&D no se responsabiliza de ninguna pérdida producida, como errores de funcionamiento o pérdida
de datos, que pueda derivarse del uso de esta báscula.
※ No se garantiza la conexión de esta báscula con todos los dispositivos compatibles con Bluetooth.

Accesorios
Si alguno de los accesorios falta o está dañado, póngase en contacto con el proveedor del producto o
con una sucursal de A&D.
Precision Health Scale

Instruction Manual
Mode d'emploi

Manuale d’uso

Italiano

Bedienungsanleitung

DeutschManual instructivo

Español Français English

UC-351PBT-Ci

1WMPD4002776

Manual de instrucciones

Español 3

4 pilas alcalinas

Nombres de las partes
BásculaCompartimento de las pilas
Asa
Interruptor de ajuste

Patas

PANTALLA
Símbolo decomunicación
Bluetooth

Símbolo de Estable
Indicador de carga de las pilas

UnidadSímbolos visualizadosSección de pantalla
Símbolo○ParpadeanteNombreDescripción
Este símbolo se visualiza cuando la indicación de la
medición es estable.
Indicador de carga normal de las pilas, conindicación de la carga restante.

Símbolo de Lista/Estable
Indicador de carga de las pilas↓
Indicador de bajo nivel de cargade las pilas

Si se visualiza este símbolo, sustituya las cuatro
pilas por unas nuevas.

Símbolo de comunicación BluetoothParte inferior de la báscula
Símbolo

NombreDescripción

Etiqueta WEEE-----

Español 4

Instalación (o sustitución) de laspilas
la tapa del compartimento de las
1. Abrapilas situado en la parte inferior de labáscula.

cuatro pilas alcalinas nuevas de
2. Instaletamaño AA alineando correctamente lospolos positivo (+) y negativo (-) con lossímbolos correspondientes indicados enel compartimento de las pilas.

la tapa del compartimento de las pilas.
3. CierreCuando se instalan las pilas, la báscula pasa al modo deconexión.
Para conectar la báscula con un receptor, consulte "Registro
con un receptor (conexion)" (P.6). Si no desea conectar labáscula con un receptor, consulte "Omisión del registro con
un receptor (conexion)" (P.7).

Precaución
Instale cada una de las pilas alineando correctamente los polos positivo (+) y negativo (-) con lossímbolos correspondientes indicados en el compartimento de las pilas.
La instalación incorrecta de las pilas podría provocar una fuga del fluido de las pilas.
Si se visualiza el símbolo, sustituya las cuatro pilas por unas nuevas.
No mezcle pilas nuevas y usadas. Si lo hace, podría provocar daños en la báscula o acortar la
duración de las pilas.
Asegúrese de utilizar pilas alcalinas de tamaño AA. No utilice pilas de ningún otro tipo.
No utilice nunca pilas recargables en esta báscula.
La duración de las pilas depende de la temperatura ambiental, el tipo de pila y el uso.
Las pilas suministradas con el dispositivo están destinadas a probar el rendimiento de la báscula ypueden tener una duración limitada.
Extraiga las pilas de su compartimento cuando la báscula no vaya a usarse durante un periodo
prolongado de tiempo. Si la báscula se almacena con las pilas instaladas, podría producirse una
fuga del fluido de las pilas, provocando daños en la báscula. Las pilas suministradas están
destinadas a comprobar el funcionamiento de la báscula y pueden tener una vida útil reducida.

Función de apagado automático
La báscula tiene una función de "Apagado automático".
Después de que los resultados de la medición se visualizan durante 15 segundos, la báscula seapaga automáticamente.
Español 5

Registro con un receptor (conexión)
Esta báscula debe registrarse con un receptor para utilizar la comunicación Bluetooth con dicho receptor.
El proceso de registro de la báscula con un receptor se conoce como conectar.
Una vez completada la conexión, los datos de medición de la báscula se envían automáticamente aldispositivo conectado cada vez que se realiza una medición, siempre que el receptor esté en un estado
que le permita recibir transmisiones.
Consulte el manual de instrucciones del receptor y ajústelo para habilitar la conexión. Si el receptor tieneun "asistente de conexión", utilícelo para realizar la conexión.
Coloque el receptor y la báscula tan cerca como sea posible para llevar a cabo la conexión.

cuatro pilas alcalinas de tamaño AA
1. Instaleen la báscula.
Para obtener más información, consulte "Instalación (osustitución) de las pilas" (P.5).

que se visualiza "Pr".
2. Confirme
" " parpadea.

≪♪Pitido≫

El receptor puede detectar esta báscula.
La conexión puede omitirse utilizando el método siguiente.
i） Pulse el interruptor de ajuste cuando se visualice "Pr".

la conexión con el receptor.
3. Inicie
Siga las instrucciones de conexión que permitan al receptor
efectuar la búsqueda, seleccionar esta báscula y conectarse con
ella.
Si el receptor solicita un código PIN, introduzca "39121440".

4. Confirme que se visualiza "End".

≪♪Dos pitidos≫

La conexión se ha completado.
Si se visualiza "E-10" o "E-11", vuelva a realizar el procedimiento
de conectar desde el paso 1.

※ Esta báscula puede conectarse con dos receptores.
※ Si la báscula se conecta con un tercer receptor, permanece la información del receptor con el que seha comunicado más recientemente.
※ Una vez que la báscula se ha conectado con el receptor, no es necesario realizar la conexión
después de sustituir las pilas.

Español 6

Precauciones respecto a la conexión
Apague el resto de los dispositivos Bluetooth cuando realice la conexión.
No es posible conectar varios dispositivos simultáneamente.
Si los datos de la medición no pueden recibirse, pruebe a realizar el emparejamiento intente laconexión de nuevo.
Una vez completada la conexión, todas las comunicaciones se establecen automáticamente.
Para desconectar un receptor, consulte "Desconectar" (P.10).
En el improbable caso de que esta báscula provocase interferencias de ondas de radio a otra
estación inalámbrica, cambie la ubicación de la báscula o deje de usarla inmediatamente.

Omisión del registro con unreceptor (conexión)
cuatro pilas alcalinas de tamaño AA
1. Instaleen la báscula.
Para obtener más información, consulte "Instalación (osustitución) de las pilas" (P.5).

el interruptor de ajuste cuando se
2. Pulse
visualice "Pr".
Pulse el interruptor de ajuste para apagar la báscula.

Español 7≪♪Pitido≫

Ajuste de la unidad
que la báscula
1. Confirme
esté apagada ymantenga pulsado elinterruptor de ajuste.

≪♪Pitido≫

Cuando el interruptor de ajuste sepulsa de forma continuada, "unit" y"uPr" se visualizan
alternativamente.

el interruptor de
2. Suelte
ajuste cuando se≪♪Pitido≫

visualice "unit".
La pantalla pasa de indicar "unit" aindicar la unidad que está
actualmente ajustada.
kg

el interruptor de
3. Pulse
ajuste para seleccionarkg

la unidad que desee
ajustar.
La unidad cambia cada vez que sepulsa el interruptor de ajuste.
Espere durante 3 segundos
después de ajustar la unidad.

lb

que se4. Confirmevisualiza "End".

≪♪Dos pitidos≫

Los ajustes se guardan.
Transcurridos tres segundos sinefectuar ninguna operación, labáscula confirma la unidad
seleccionada con dos pitidos,
indica "End" y se apaga
automáticamente.

Español 8

Medición del peso
Subirse a la báscula
que la báscula
1. Confirme
esté apagaday súbase aella con cuidado.

La medición se inicia cuando usted
se sube a la báscula.
Durante la medición se visualiza una≪♪Pitido≫

barra de progreso. Permanezca
erguido y tan quieto como sea
posible mientras se realiza la
medición.

de la báscula
2. Bájese
cuando "StEP" y "oFF" se≪♪ Tres pitidos≫

visualicen alternativamente
con tres pitidos.
Si no se baja, la báscula emitirá
pitidos cada dos segundos.
La báscula se apaga
automáticamente transcurridos 2
minutos.
Cuando se baje de la báscula,aparecerán encendidos todos lossegmentos del display.
No toque la báscula mientras
aparecen encendidos todos lossegmentos del display.
Si lo hace, podría producirse unerror de lectura.

in
c
lb
kg

el resultado de
3. Confirme
la medición.

kgLa báscula se apaga
automáticamente transcurridos 15
segundos desde la visualizacióndel resultado de la medición.
Los datos se envían
automáticamente al receptor siéste se encuentra en un estado
que le permita recibirlos.
Si la báscula no está conectada
con un receptor, los datos sealmacenan en la memoria de labáscula.
(Consulte "Transmisión de datos
almacenados temporalmente"(P.10).)
Los datos almacenados en lamemoria de la báscula no pueden
visualizarse en esta báscula.
También puede apagar la básculapulsando el interruptor de ajuste.
Español 9

Otras funciones
Desconexión
que la báscula
1. Confirme
esté apagada ymantenga pulsado elinterruptor de ajuste.
Cuando el interruptor de ajuste sepulsa de forma continuada, "unit" y"uPr" se visualizan
alternativamente.

2. Suelte el interruptor de≪♪Pitido≫≪♪Pitido≫

ajuste cuando se
visualice "uPr".

que se3. Confirmevisualiza "End".

≪♪Dos pitidos≫

El receptor se ha desemparejado.
Si se visualiza "E-10" o "E11", vuelva
a realizar el procedimiento dedesemparejamiento desde el paso 1.
La báscula se apaga
automáticamente.

Transmisión de datos almacenados temporalmente
Si los datos de la medición no pueden enviarse a un receptor, la báscula puede almacenar
temporalmente los datos de hasta 200 mediciones en su memoria integrada. Si la báscula puedetransmitir a un receptor, se transmiten todos los datos. Si el volumen de datos temporalmente
almacenados supera las 200 mediciones, los datos más antiguos se eliminan y se sustituyen por datos
nuevos.
※ Los datos almacenados no pueden visualizarse en esta báscula.
※ El volumen máximo de datos que pueden almacenarse temporalmente puede variar en función delreceptor.
Español 10

Comunicación de la báscula
Esta báscula está equipada con una función de comunicación inalámbrica Bluetooth certificada por
Continua. Los datos de medición pueden transmitirse conectándose a un dispositivo Bluetooth que sea
compatible con las especificaciones de conexión de Continua.
Los dispositivos certificados por Continua llevan la marca del logotipo de Continua.
Los dispositivos Bluetooth llevan la marca del logotipo de Bluetooth.
Para obtener más información sobre cómo conectar esta báscula con un receptor equipado con
Bluetooth, consulte "Registro con un receptor (Conexión)" (P.6).
Consulte también el manual de instrucciones de conexión del receptor con el que desee conectar labáscula.

Pautas para garantizar una buena conexión
Asegúrese de que no existan obstáculos en el espacio existente entre la báscula y el receptor. La
distancia de comunicación podría acortarse debido a la estructura del edificio u otros obstáculos. En
particular, la comunicación podría no ser posible a través de paredes de hormigón armado, etc.
No utilice la conexión Bluetooth cerca de una LAN inalámbrica u otros equipos inalámbricos, cerca dedispositivos que emitan ondas electromagnéticas (por ejemplo, hornos microondas), en lugares
donde exista un gran número de obstáculos ni en ningún otro tipo de entorno inadecuado para las
ondas de radio. Ello podría provocar que la conexión se interrumpiese con frecuencia, que lavelocidad de comunicación disminuyese considerablemente o que se produjeran errores.
Cuando se utilizan cerca de un dispositivo de LAN inalámbrica IEEE802.11g/b/n, los dispositivos
provocan interferencias mutuas de ondas de radio que pueden disminuir la velocidad decomunicación o imposibilitar la conexión. Si esto sucede, apague el dispositivo que no esté utilizando
o cambie el lugar donde se esté utilizando.
Si la báscula no se comunica correctamente al usarse cerca de un dispositivo inalámbrico o una
estación de radiodifusión, cambie el lugar donde se esté utilizando.

PrecauciónLa distancia de comunicación entre la báscula y el receptor depende de la clase de salida
Bluetooth del receptor.
Para receptores Bluetooth de clase 1: Máx. 100 m
Para receptores Bluetooth de clase 2: Máx. 10 m
Esta distancia podría acortarse a causa del entorno circundante; por tanto, compruebe que losdispositivos estén lo suficientemente cerca para que los datos puedan transmitirse tras la
medición.

Ajustes de fecha y hora
Esta báscula tiene un reloj integrado. La fecha y la hora de la medición se incluyen en los datos de la
medición. El reloj está sincronizado con el reloj del receptor. Para obtener más información sobre las
especificaciones del reloj, consulte el manual de instrucciones del receptor.
Español 11

Especificaciones
ModeloUC-351PBT-Ci

Capacidad máxima

150 kg / 330 lb

Indicación mínima

0,1 unidades de kg de 0 a 100,0 kg
0,2 unidades de kg de 100,0 a 150,0 kg
0,2 unidades de lb de 0 a 220,0 lb
0,5 unidades de lb de 220,0 a 330,0 lb

Comunicación
inalámbrica

WML-C40AH (MITSUMI ELECTRIC CO., LTD.)
Bluetooth Ver. 2.1 clase 1
Compatible con SSP y HDP
Banda de frecuencias: de 2402 MHz a 2480 MHz
Potencia de salida de RF máxima: 20 dBm

Opciones de ajuste

Unidades: kg o lb

Fuente dealimentación

Cuatro pilas alcalinas de tamaño AA

Duración de las pilas

Aprox. 1 año (dos mediciones al día si se usan pilas alcalinas)

Dimensiones

350 (anchura) ×350 (profundidad) ×39 (altura) mm

Peso

Aprox. 2,3 kg (incluyendo pilas)

Accesorios

Manual de instrucciones, pilas alcalinas

Conformidad con las directivas europeas
La báscula cumple con la directiva legal EMC (compatibilidad electromagnética)
2014/30/UE.
Por la presente, MITSUMI ELECTRIC CO., LTD. declara que el tipo de equipo radioeléctrico
WML-C40AH es conforme con la Directiva 2014/53/UE. El texto completo de la declaración
UE de conformidad está disponible en la dirección Internet siguiente:
http://www.aandd.jp/products/manual/manual_medical.html.
Notas: Solo para uso en interiores. El valor indicado puede verse negativamente afectado por la
exposición a influencias electromagnéticas extremas.

Conformidad con las normas de la FCC
La báscula cumple con el apartado 15 de las normas de la FCC y contiene el identificador
POOWML-C40 de la FCC.


Conformidad con el Ministerio de Industria de Canadá
IC: 4250A-WMLC40.

Español 12

Antes de solicitar asistencia técnica
Síntomas

Medida correctiva

Se visualiza E

Supera la capacidad máxima (150 kg / 330 lb).

Se visualiza E-0

Supera la capacidad máxima (150 kg / 330 lb).

Se visualiza Err

Supera la capacidad máxima (150 kg / 330 lb).

Se visualiza E-10

La conexión o desconexión no se ha completado correctamente.
Extraiga y vuelva a instalar las pilas.

Se visualiza E-11
Se visualiza H
Se visualiza
No se visualiza nada alpulsar el interruptor deajusteSupera la capacidad máxima (150 kg / 330 lb).
Extraiga y vuelva a instalar las pilas. O bien, sustituya las cuatro pilas por
pilas alcalinas nuevas de tamaño AA .

El resultado
de la medición difieresignificativamente delvalor esperado.

Coloque la báscula sobre una superficie sólida nivelada.

La medición no se inicia
cuando me subo a labáscula.

¿Ha pulsado el interruptor de ajuste al subirse a la báscula?
Coloque la báscula sobre un suelo duro y pulse el interruptor de ajuste
antes de realizar la medición.

Retire cualquier objeto extraño que pueda estar tocando la parte inferior
de la báscula.

Mantenimiento de la básculaSi la báscula se ensucia, empape un paño en agua o detergente suave, escúrralo bien y limpie con élla báscula.
Después de limpiar la suciedad con el paño húmedo, seque la báscula con un paño seco y suave.
No utilice nunca alcohol, benceno, diluyentes o productos químicos agresivos para limpiar la báscula.
Extraiga las pilas de su compartimento cuando la báscula no vaya a usarse durante un periodo
prolongado de tiempo.

Español 13

Indice
Guida all’accensione ................................................................................. 1
Precauzioni di sicurezza ........................................................................... 2
Funzione di comunicazione senza fili...................................................... 3
Accessori.................................................................................................... 3
Descrizione delle parti............................................................................... 4
Installazione (o sostituzione) delle batterie ............................................ 5
Registrazione con un ricevitore (associazione) ..................................... 6
Come saltare la registrazione con un ricevitore (associazione) ........... 7
Impostazione dell’unità ............................................................................. 8
Misurazione del proprio peso ................................................................... 9
Altre funzioni ............................................................................................ 10
Caratteristiche tecniche .......................................................................... 12
Prima di richiedere un intervento di assistenza ................................... 13
Manutenzione della bilancia ................................................................... 13
Gentili clienti
Congratulazioni per aver acquistato la bilancia di precisione A&D, attualmente una delle bilance piùavanzate. Questo apparecchio, progettato per offrire facilità d’uso e precisione, è un ausilio per leoperazioni quotidiane di rilevazione del peso. Prima di utilizzare questo apparecchio per la prima volta,
si consiglia di leggere completamente e con attenzione il presente manuale.

1WMPD4002776A

Guida all’accensione
Operazioni

Installare le batterie (P.5)

Eseguire l’associazione con un ricevitore (P.6)
Eseguire la misurazione senza l’associazione con un ricevitore(P.7)

Impostare l’unità (P.8)

Misurazione(P.9)

Salire sulla bilancia

Verificare che le indicazioni StEP e oFF vengano visualizzate inmodo alternato

Scendere dalla bilancia

Verificare il proprio pesokg

※ I dati vengono trasmessi.
I dati vengono memorizzati nella bilancia, qualora non sia presente alcun
ricevitore. (P.10)● È anche possibile rimuovere l’associazione con i ricevitori. (P.10)

Italiano 1

Precauzioni di sicurezza
Le voci seguenti contengono informazioni per evitare il rischio di causare danni o lesioni alla propria o ad altre
persone. Leggere queste voci con attenzione e assicurarsi di utilizzare correttamente il presente prodotto.

Pericolo

Questo simbolo indica “un pericolo immediato che potrebbe
causare il decesso o gravi lesioni personali”.

Attenzione

Questo simbolo indica “la possibilità di decesso o di lesione
personale grave”.
Questo simbolo indica “la possibilità di lesione fisica o danni

Avvertenza alle cose”.

Precauzioni per l’usoPrecauzioni generali

Attenzione
Non affidarsi all’autodiagnosi nel tentativo di perdere peso o di svolgere attività fisica terapeutica.
Attenersi alle istruzioni di un medico o di uno specialista.
Non salire sui bordi della bilancia né saltare su quest’ultima.
In caso contrario, si potrebbe provocare una caduta, causando lesioni personali. L’impatto potrebbe anchedanneggiare la bilancia.
Non utilizzare la bilancia su superfici irregolari.
In caso contrario, si potrebbe danneggiare la bilancia. Inoltre, la bilancia non sarebbe in grado di eseguire una
misurazione accurata.
Non utilizzare la bilancia su superfici scivolose, né salire sulla bilancia con i piedi bagnati.
In caso contrario, si potrebbe provocare uno scivolamento e causare lesioni personali. Inoltre, la bilancia nonsarebbe in grado di eseguire una misurazione accurata.

AvvertenzaNon collocare la bilancia in ambienti in cui sia esposta ad acqua o umidità eccessive, ad esempio in un bagno; inambienti in cui sia esposta alla luce solare diretta; in ambienti in cui sia esposta direttamente all’apertura di uscita
dell’aria di un condizionatore d’aria o in prossimità del fuoco.
In caso contrario, si potrebbe causare il malfunzionamento della bilancia.
La presente bilancia va utilizzata esclusivamente per la misurazione del peso corporeo.
La presente bilancia è uno strumento di misurazione di precisione. Non far cadere la bilancia, né esporla a vibrazioni
o a forti impatti.
Non smontare, riparare o modificare la bilancia.
Queste operazioni potrebbero causare un malfunzionamento o lesioni personali.
I disabili dovrebbero essere aiutati da una persona adibita alla loro assistenza, durante l’uso della presente bilancia.
In caso contrario, si potrebbe provocare una caduta, causando lesioni personali.
Quando si installano le batterie, assicurarsi che i poli + (positivo) e - (negativo) siano orientati correttamente.
Un’installazione errata potrebbe provocare un surriscaldamento, perdite di fluido elettrolitico dalle batterie o la rottura
di queste ultime, causando il danneggiamento della bilancia o lesioni personali.
La presente bilancia è stata progettata per l’uso personale.
Riporre la presente bilancia al di fuori della portata dei bambini piccoli.
Non riporre la presente bilancia in posizione verticale. Qualora, per qualsiasi motivo, la presente bilancia venga
riposta in posizione verticale, assicurarsi di fare in modo che non cada.
Non collocare alcun oggetto sulla bilancia mentre è riposta.
Qualora l’utilizzo della presente bilancia venga condiviso con una persona che soffra di una malattia infettiva, ad
esempio una malattia della pelle sui piedi, e così via, pulire la bilancia prima dell’uso.
(Per i dettagli sulla pulizia della bilancia, vedere “Manutenzione della bilancia (P.13)”.)
Per misurare accuratamente il proprio peso, utilizzare la bilancia su un pavimento piatto e rigido. Non utilizzarla su
un tappeto o un’altra superficie morbida.

Italiano 2

fili

Funzione di comunicazione senza

La presente bilancia utilizza l’interfaccia senza fili Bluetooth per comunicare con un apparecchio
ricevente. Alla fine della misurazione, i dati della misurazione vengono inviati automaticamente dallapresente bilancia all’apparecchio ricevente.
Il Bluetooth è una tecnologia di comunicazione che consente agli apparecchi di comunicare senza fili.
Per i dettagli sul Bluetooth, vedere “Comunicazione con la bilancia (P.11)”.

Apparecchio ricevente
Si tratta di un apparecchio quale, ad esempio, un personal computer o uno smartphone, sul quale sia
stata scaricata e installata un’applicazione per consentire la comunicazione via Bluetooth con lapresente bilancia.

Precauzioni per la funzione di trasmissione senza fili
Attenzione
Non utilizzare questa funzione in luoghi in cui le comunicazioni senza fili siano vietate, ad esempio sugli aerei o negli
ospedali.
La presente bilancia potrebbe produrre effetti negativi su apparecchi elettronici o apparecchiature elettromedicali.

Avvertenza
Qualora la funzione di comunicazione della presente bilancia venga utilizzata in prossimità di un apparecchio senzafili che comunichi sulla gamma dei 2,4 GHz, la velocità di elaborazione di entrambi gli apparecchi potrebbe ridursi.
Non utilizzare in prossimità di un forno a microonde, dove potrebbero verificarsi interferenze dovute a campi
magnetici, elettricità statica o onde radio (il segnale potrebbe non pervenire a causa di motivi ambientali.)※ A&D non si assume alcuna responsabilità per eventuali perdite dovute a malfunzionamenti durante
l’uso o perdite di dati che potrebbero verificarsi attraverso l’uso della presente bilancia.
※ Non è garantita la connessione della presente bilancia a tutti gli apparecchi compatibili Bluetooth.

Accessori
Qualora un qualsiasi accessorio risulti mancante o danneggiato, contattare il produttore o una filiale divendita A&D.
Precision Health Scale

Instruction Manual
Mode d'emploi

Manuale d’uso

Italiano

Bedienungsanleitung

DeutschManual instructivo

Español Français English

UC-351PBT-Ci

1WMPD4002776

Manuale d’uso

Batterie alcaline x 4

Italiano 3

Descrizione delle parti
Bilancia
Vano batterie
Maniglia
Interruttore di impostazione

PiediniSezione del display
Simbolo della
comunicazione
Bluetooth

Simbolo Stabile
Indicatore delle batterie

Unità

Simboli sul display
Sezione del display
Simbolo○

Lampeggiant

Nome

Descrizione

Questo simbolo viene visualizzato quando
l’indicazione della misurazione è stabile.
Il normale indicatore di carica delle batterie con la
Indicatore delle batterie
carica residua.
↓
Qualora venga visualizzato questo simbolo,
Indicatore di batterie scariche
sostituire tutte e quattro le batterie con batterienuove.
Simbolo della comunicazione Bluetooth
Simbolo Pronta/Stabile
Lato inferiore della bilancia
Simbolo

Nome

Descrizione

Etichetta RAEE-----

Italiano 4

Installazione (o sostituzione) dellebatterie
il coperchio del vano batterie sul
1. Aprire
lato inferiore della bilancia.

quattro nuove batterie alcaline
2. Installare
formato AA con i poli positivi (+) e
negativi (-) allineati correttamente con isimboli corrispondenti nel vano batterie.
il coperchio del vano batterie.
3. ChiudereLa bilancia passa alla modalità di associazione quando si
installano le batterie.
Per associare la bilancia con un ricevitore, vedere“Registrazione con un ricevitore (associazione) (P.6)”.
Qualora non si desideri associare la bilancia con un
ricevitore, vedere “Come saltare la registrazione con un
ricevitore (associazione) (P.7)”.

Avvertenza
Installare ciascuna batteria con i poli positivo (+) e negativo (-) allineati correttamente con i simboli
corrispondenti nel vano batterie.
Un’installazione errata delle batterie potrebbe provocare la fuoriuscita di liquido elettrolitico delle batterie.
viene visualizzato.
Sostituire tutte e quattro le batterie con batterie nuove, quando il simbolo
Non mescolare batterie nuove con batterie usate. In caso contrario, si potrebbe danneggiare labilancia o ridurre la durata delle batterie.
Assicurarsi di utilizzare batterie alcaline in formato AA. Non utilizzare altri tipi di batterie.
Non utilizzare mai batterie ricaricabili con la presente bilancia.
La durata delle batterie dipende dalla temperatura ambiente, dal tipo di batterie e dall’utilizzo.
Le batterie fornite con l’apparecchio servono per verificare le prestazioni della bilancia, epotrebbero avere una durata limitata.
Quando non si intende utilizzare la bilancia per un periodo di tempo prolungato, rimuovere le batterie dalvano batterie. Qualora si riponga la bilancia con le batterie installate, si potrebbe causare la fuoriuscita diliquido elettrolitico dalle batterie, provocando il danneggiamento della bilancia. Le batterie fornite servono
per controllare il funzionamento della bilancia, e potrebbero avere una durata breve.

Funzione di spegnimento automatico
La bilancia dispone di una funzione di “Spegnimento automatico”.
Dopo che i risultati della misurazione sono stati visualizzati per 15 secondi, la bilancia si spegne
automaticamente.
Italiano 5

Registrazione con un ricevitore(associazione)
La presente bilancia deve essere associata con un ricevitore, per poter utilizzare la comunicazione
Bluetooth con il ricevitore. La procedura di registrazione della presente bilancia con un ricevitore viene
chiamata associazione.
Al termine dell’associazione, i dati di misurazione provenienti dalla presente bilancia vengono inviati
automaticamente all’apparecchio associato ogni volta che si esegue una misurazione, quando il
ricevitore è in uno stato in cui può ricevere le trasmissioni.
Consultare il manuale d’uso del ricevitore e impostarlo in modo da consentire l’associazione. Qualora il
ricevitore disponga di una “procedura guidata di associazione”, utilizzare quest’ultima per completarel’associazione.
Posizionare il ricevitore e la presente bilancia alla minima distanza reciproca possibile per eseguirel’associazione.

quattro batterie alcaline formato
1. Installare
AA nella presente bilancia.
Per i dettagli, vedere “Installazione (o sostituzione) delle batterie(P.5)”.

che venga visualizzata
2. Verificare
l’indicazione “Pr”.

≪♪Bip≫

Il simbolo “ ” lampeggia.
La bilancia è rilevabile dal ricevitore.
È possibile saltare l’associazione utilizzando il metodo
seguente.
i） Premere l’interruttore di impostazione quando viene
visualizzata l’indicazione “Pr”.

l’associazione sul ricevitore.
3. Avviare
Seguire le istruzioni di associazione per il ricevitore per eseguirela ricerca, la selezione e l’associazione con la presente bilancia.
Qualora il ricevitore richieda un codice PIN, immettere“39121440”.

che l’indicazione “End” venga
4. Verificare
visualizzata.

≪♪Bip Bip≫

Associazione completata.
Qualora venga visualizzata l’indicazione “E-10” o “E-11”,
eseguire di nuovo la procedura di associazione dal punto 1.
※ È possibile associare la presente bilancia con due ricevitori.
※ Qualora la presente bilancia venga associata con un terzo ricevitore, vengono conservate leinformazioni del ricevitore con cui si è verificata la comunicazione più di recente.
※ Una volta che la bilancia è stata associata con il ricevitore, non è necessario eseguire l’associazione
dopo aver sostituito le batterie.

Italiano 6

Avvertenze per l’associazione
Quando si intende eseguire l’associazione, spegnere gli altri apparecchi Bluetooth.
Non è possibile associare più apparecchi contemporaneamente.
Qualora non si riesca a ricevere i dati delle misurazioni, provare a eseguire di nuovol’associazione.
Dopo aver completato l’associazione, tutte le comunicazioni vengono effettuate automaticamente.
Per rimuovere l’associazione con un ricevitore, vedere “Rimozione dell’associazione (P.10)”.
Nella remota eventualità che la presente bilancia provochi interferenze relative alle onde radio nei
confronti di un’altra stazione di comunicazione senza fili, cambiare l’ubicazione in cui viene
utilizzata la bilancia o interrompere immediatamente l’uso della bilancia.

Come saltare la registrazione con
un ricevitore (associazione)
quattro batterie alcaline formato
1. Installare
AA nella presente bilancia.
Per i dettagli, vedere “Installazione (o sostituzione) delle batterie(P.5)”.

l’interruttore di impostazione
2. Premere
quando viene visualizzata l’indicazione“Pr”.
Premere l’interruttore di impostazione per spegnere
l’apparecchio.

Italiano 7≪♪Bip≫

Impostazione dell’unità
che la bilancia
1. Verificare
sia spenta, quindi tenere
premuto l’interruttore di
impostazione.

≪♪Bip≫

Quando si tiene premuto
l’interruttore di impostazione
continuativamente, le indicazioni“unit” e “uPr” vengono visualizzate
in modo alternato.

l’interruttore
2. Rilasciare
di impostazione quando≪♪Bip≫

viene visualizzata
l’indicazione “unit”.
L’indicazione visualizzata cambia
da “unit” alla visualizzazione
dell’unità correntemente impostata.

kg

l’interruttore di
3. Premere
impostazione perkg

selezionare l’unità che si
desidera impostare.
L’unità cambia ad ogni pressionedell’interruttore di impostazione.
Attendere tre secondi dopo aver
impostato l’unità.

lbche
4. Verificare
l’indicazione “End”≪♪Bip Bip≫

venga visualizzata.
Le impostazioni vengono salvate.
Dopo tre secondi senza effettuare
operazioni, la bilancia conferma
l’unità selezionata con due “bip”,
visualizza l’indicazione “End”,
quindi si spegne automaticamente.

Italiano 8

Misurazione del proprio peso
Misurazione del peso salendo sulla bilancia
che la bilancia sia
1. Verificare
spenta, quindi salire

delicatamente sulla bilancia.
La misurazione si avvia quando si
sale sulla bilancia.
Durante la misurazione, viene≪♪Bip≫

visualizzata una barra diavanzamento. Stare in piedi senza
piegarsi e rimanere immobili il più
possibile durante la misurazione.

dalla bilancia quando
2. Scendere
le indicazioni “StEP” e “oFF”≪♪Bip bip bip≫

vengono visualizzate in modo
alternato con tre “bip”.

Se non si scende dalla bilancia,
quest’ultima emette un “bip” ogni
due secondi.
La bilancia si spegne
automaticamente dopo 2 minuti.
Tutti i segmenti del display
vengono visualizzati quando si
scende dalla bilancia.
Non toccare la bilancia mentre tutti
i segmenti del display sonovisualizzati.
In caso contrario, si potrebbe
provocare una lettura errata.

in
c
lb
kg

i risultati della
3. Verificare
misurazione.

kg

La bilancia si spegne
automaticamente 15 secondi dopo
aver visualizzato i risultati dellamisurazione.
I dati vengono inviati
automaticamente al ricevitore, sequest’ultimo è nello stato diricezione.
Qualora la bilancia non sia
associata con un ricevitore, i dati
vengono archiviati nella memoriadella bilancia.
(Vedere “Trasmissione dei dati
memorizzati temporaneamente(P.10)”.)
Non è possibile visualizzare sullapresente bilancia i dati archiviati
nella memoria della bilancia stessa.
È anche possibile spegnere
l’apparecchio premendo
l’interruttore di impostazione.
Italiano 9

Altre funzioni
Rimozione dell’associazione
che la bilancia
1. Verificare
sia spenta, quindi tenere
premuto l’interruttore di
impostazione.

Quando si tiene premuto
l’interruttore di impostazione
continuativamente, le indicazioni“unit” e “uPr” vengono visualizzate
in modo alternato.

≪♪Bip≫

l’interruttore
2. Rilasciare
di impostazione quando≪♪Bip≫

viene visualizzata
l’indicazione “uPr”.

che
3. Verificare
l’indicazione “End”≪♪Bip Bip≫

venga visualizzata.

L’associazione con il ricevitore
viene rimossa.
Qualora venga visualizzata
l’indicazione “E-10” o “E-11”, eseguiredi nuovo la procedura di rimozione
dell’associazione dal punto 1.
La bilancia si spegne
automaticamente.

Trasmissione dei dati memorizzati temporaneamente
Qualora non sia possibile inviare i dati delle misurazioni a un ricevitore, la presente bilancia è in grado diarchiviare temporaneamente i dati di un massimo di 200 misurazioni nella memoria incorporata. Quando
la bilancia è in grado di trasmettere a un ricevitore, tutti i dati vengono inviati. Qualora la quantità di dati
memorizzati temporaneamente superi le 200 misurazioni, i dati più vecchi vengono eliminati e sostituiti
con dati nuovi.
※ Non è possibile visualizzare i dati memorizzati sulla presente bilancia.
※ La quantità massima di dati che possono essere memorizzati temporaneamente può variare aseconda del ricevitore.
Italiano 10

Comunicazione con la bilancia
La presente bilancia è dotata di una funzione di comunicazione senza fili Bluetooth con certificazione
Continua. È possibile trasmettere i dati delle misurazioni mediante la connessione a un apparecchio
Bluetooth che sia compatibile con le specifiche di connessione Continua.
Gli apparecchi dotati di certificazione Continua recano il logo Continua.
Gli apparecchi Bluetooth recano il logo Bluetooth.
Per i dettagli sull’associazione della presente bilancia con un ricevitore che sia dotato di Bluetooth,vedere “Registrazione con un ricevitore (associazione) (P.6)”.
Fare riferimento anche al manuale d’uso per l’associazione del ricevitore con cui si desidera
effettuare l’associazione.

Come assicurare una buona connessione
Assicurarsi che lo spazio tra la presente bilancia e il ricevitore sia privo di ostacoli. La distanza di
comunicazione potrebbe ridursi a causa della struttura dell’edificio o di altri ostacoli. In particolare, lacomunicazione potrebbe non essere possibile attraverso pareti in cemento armato, e così via.
Non utilizzare la connessione Bluetooth in prossimità di una LAN wireless o di altri apparecchi senzafili, in prossimità di apparecchi che emettano onde elettromagnetiche, ad esempio forni a microonde,in ubicazioni che presentino un numero elevato di ostacoli o in altri ambienti inadatti alle onde radio.
In caso contrario, si potrebbero causare interruzioni frequenti della connessione, riduzioni sostanziali
della velocità di comunicazioni ed errori.
Se utilizzati in prossimità di un apparecchio LAN wireless IEEE802.11g/b/n, i presenti apparecchi
causano interferenze reciproche relative alle onde radio, e la velocità di comunicazione potrebbe
ridursi o la connessione potrebbe risultare impossibile. In situazioni di questo tipo, spegnere
l’apparecchio che non si sta utilizzando, oppure cambiare l’ubicazione in cui viene utilizzato.
Qualora la bilancia non comunichi correttamente quando viene utilizzata in prossimità di unapparecchio senza fili o di una stazione di trasmissione, cambiare l’ubicazione in cui viene utilizzata.

AvvertenzaLa distanza di comunicazione tra la presente bilancia e il ricevitore dipende dalla classe di potenza
di trasmissione Bluetooth del ricevitore.
Per i ricevitori Bluetooth di Classe 1: Max. 100 m
Per i ricevitori Bluetooth di Classe 2: Max. 10 m
Questa distanza potrebbe ridursi a causa dell’ambiente circostante; pertanto, controllare che ladistanza sia sufficientemente ravvicinata da consentire la trasmissione dei dati dopo lamisurazione.

Impostazioni dell’ora
La presente bilancia dispone di un orologio incorporato. La data e l’ora della misurazione sono inclusenei dati di misurazione. L’orologio viene sincronizzato con quello presente nel ricevitore. Per i dettagli
sulle caratteristiche tecniche dell’orologio, vedere il manuale d’uso del ricevitore.
Italiano 11

Caratteristiche tecniche
Modello

UC-351PBT-Ci

Portata massima

150 kg / 330 lb

Indicazione minima
visualizzata

Unità di 0,1 kg da 0 a 100,0 kg
Unità di 0,2 kg da 100,0 a 150,0 kg
Unità di 0,2 lb da 0 a 220,0 lb
Unità di 0,5 lb da 220,0 a 330,0 lb

Comunicazione
senza fili

WML-C40AH (MITSUMI ELECTRIC CO., LTD.)
Bluetooth Ver.2.1 classe 1
Compatibile SSP HDP
Banda di frequenza: da 2402 MHz a 2480 MHz
Potenza max in uscita di RF: 20 dBm

Voci di impostazione

Unità: kg o lb

Alimentazione

Quattro batterie alcaline formato AA

Durata delle batterie

Circa 1 anno (con due misurazioni al giorno, se si utilizzano batterie alcaline)

Dimensioni

350 (L) ×350 (P) ×39 (A) mm

Peso

Circa 2,3 kg (batterie incluse)

Accessori

Manuale d’uso, batterie alcalineConformità alle Direttive Europee
La bilancia è conforme alla direttiva obbligatoria EMC (compatibilità elettromagnetica)
2014/30/UE.
Il fabbricante, MITSUMI ELECTRIC CO., LTD., dichiara che il tipo di apparecchiatura radio
WML-C40AH è conforme alla direttiva 2014/53/UE. Il testo completo della dichiarazione diconformità UE è disponibile al seguente indirizzo Internet:
http://www.aandd.jp/products/manual/manual_medical.html
Note: solo per uso in interni. Il valore visualizzato potrebbe essere compromesso dalla presenzadi campi elettromagnetici di estrema entità.


Conformità alle Norme FCC
La bilancia è conforme alla sezione 15 delle Norme FCC e contiene l’FCC ID POOWML-C40.


Conformità a Industry Canada
IC: 4250A-WMLC40.

Italiano 12

Prima di richiedere un intervento di
assistenza
SintomiMisura correttiva

Viene visualizzata
l’indicazione “E”

È stata superata la portata massima (150 kg/330 lb).

Viene visualizzata
l’indicazione “E-0”È stata superata la portata massima (150 kg/330 lb).

Viene visualizzata
l’indicazione “Err”

È stata superata la portata massima (150 kg/330 lb).

Viene visualizzata
l’indicazione “E-10”

L’associazione o la rimozione dell’associazione non sono statecompletate correttamente. Rimuovere e reinstallare le batterie.

Viene visualizzata
l’indicazione “E-11”
Viene visualizzata
l’indicazione “H”

È stata superata la portata massima (150 kg/330 lb).

Viene visualizzato

Rimuovere e reinstallare le batterie. Oppure, sostituire tutte e quattro lebatterie con nuove batterie alcaline formato AA.

Non viene visualizzato
alcunché quando si premel’interruttore di
impostazione
I risultati della misurazione
sono notevolmente diversi
dai valori previsti.

Collocare la bilancia su una superficie piana solida.

La misurazione non si
avvia quando si sale sullabilancia.

Si è premuto l’interruttore di impostazione quando si è saliti sullabilancia?
Collocare la bilancia su un pavimento rigido, quindi premere l’interruttore
di impostazione prima di eseguire la misurazione.

Rimuovere eventuali oggetti estranei che potrebbero essere in contatto
con la parte inferiore della bilancia.

Manutenzione della bilancia
Qualora la bilancia si sporchi, bagnare un panno in acqua o detergente delicato, strizzarlo a fondo,
quindi passarlo sulla bilancia per pulirla.
Dopo aver pulito lo sporco con il panno umido, asciugare la bilancia utilizzando un panno morbido
asciutto.
Non utilizzare mai alcool, benzene, solventi o altri prodotti chimici aggressivi per pulire la bilancia.
Quando non si intende utilizzare la bilancia per un periodo di tempo prolungato, rimuovere le batterie
dall’apposito vano.
Italiano 13

Inhaltsverzeichnis
Erste Schritte.............................................................................................. 1
Vorsichtshinweise...................................................................................... 2
Funktion zur drahtlosen Kommunikation................................................ 3
Zubehör....................................................................................................... 3
Bezeichnung der Teile ............................................................................... 4Einsetzen (oder Austauschen) der Batterien .......................................... 5
Kopplung mit einem Empfänger............................................................... 6Überspringen der Kopplung mit einem Empfänger ............................... 7
Einstellen des Geräts ................................................................................ 8
Messen des Gewichts................................................................................ 9
Sonstige Funktionen ............................................................................... 10
Technische Daten .................................................................................... 12
Bevor Sie sich an den Kundendienst wenden ...................................... 13
Wartung der Waage ................................................................................. 13

Sehr geehrter Kunde
herzlichen Glückwunsch zum Kauf Ihrer modernen Personenwaage von A&D, einer der
fortschrittlichsten Personenwaagen auf dem Markt. Dieses Gerät, das für eine einfache Bedienung
und höchste Genauigkeit entwickelt wurde, wird Ihnen die tägliche Überprüfung Ihres Gewichts
erleichtern. Wir empfehlen Ihnen, diese Anleitung sorgfältig durchzulesen, bevor Sie das Gerät zum
ersten Mal verwenden.

1WMPD4002776A

Erste Schritte
Erste Schritte

Setzen Sie die Batterien ein (S. 5)

Koppeln Sie die Waage mit einem Empfänger (S. 6)
Führen Sie das Wiegen ohne die Kopplung mit einem Empfänger
durch (S. 7)

Stellen Sie das Gerät ein (S. 8)

Wiegen(S. 9)

Steigen Sie auf die Waage

Stellen Sie sicher, dass StEP und oFF abwechselnd angezeigt
werdenSteigen Sie von der Waage herunterÜberprüfen Sie Ihr Gewicht
kg※ Die Daten werden übertragen.
Die Daten werden in der Waage gespeichert, wenn kein Empfänger vorhanden ist.
(S. 10)● Die Kopplung mit den Empfängern kann auch aufgehoben werden. (S. 10)

Deutsch 1

Vorsichtshinweise
Die folgenden Punkte enthalten Informationen, um das Risiko von Schäden oder Verletzungen bei Ihnen
oder anderen Personen zu reduzieren. Lesen Sie diese Punkte sorgfältig durch und achten Sie darauf,
dass Sie das Produkt korrekt verwenden.

GefahrDieses Symbol verweist auf „eine unmittelbare Gefahr, die zum
Tod oder zu ernsten Verletzungen führen kann“.
Dieses Symbol verweist auf „die Möglichkeit des Todes oder

Warnung ernster Verletzungen“.
Achtung

Dieses Symbol verweist auf „die Möglichkeit physischer
Verletzungen oder von Sachschäden“.

Vorsichtsmaßnahmen bei der VerwendungAllgemeine Vorsichtsmaßnahmen

Warnung
Vertrauen Sie nicht auf eine Selbstdiagnose, wenn Sie Gewicht verlieren oder therapeutisch Sport betreibenmöchten.
Befolgen Sie die Anweisungen eines Arztes oder Spezialisten.
Treten Sie nicht auf die Kanten der Waage und springen Sie nicht auf die Waage.
Andernfalls kann es zu einem Sturz und Verletzungen kommen. Außerdem kann der Aufprall Schäden an der Waage
verursachen.
Verwenden Sie die Waage nicht auf einer unebenen Oberfläche.
Andernfalls kann es zu Schäden an der Waage kommen. Außerdem kann die Waage in diesem Fall keine genaue
Messung durchführen.
Verwenden Sie die Waage nicht auf einer rutschigen Oberfläche und treten Sie nicht mit nassen Füßen auf die
Waage.
Andernfalls könnten Sie ausrutschen und sich verletzen. Außerdem kann die Waage in diesem Fall keine genaue
Messung durchführen.

Achtung
Platzieren Sie die Waage nicht an einem Ort, an dem sie übermäßig viel Wasser oder Luftfeuchtigkeit ausgesetzt ist,
wie in einem Badezimmer, an dem sie direktem Sonnenlicht ausgesetzt ist, an dem sie dem direkten Luftstrom einer
Klimaanlage ausgesetzt ist, oder in der Nähe von Feuer.
Andernfalls kann es zu einer Fehlfunktion der Waage kommen.
Diese Waage sollte nur zum Wiegen des Körpergewichts verwendet werden.
Bei dieser Waage handelt es sich um ein Präzisionsmessgerät. Lassen Sie die Waage nicht fallen und setzen Sie sie
keinen Vibrationen oder starken Stößen aus.
Bauen Sie die Waage nicht auseinander, reparieren Sie sie nicht und verändern Sie sie nicht.
Andernfalls kann es zu einer Fehlfunktion oder zu Verletzungen kommen.
Personen mit Behinderungen sollten sich bei der Verwendung der Waage von einer Pflegeperson helfen lassen.
Andernfalls kann es zu einem Sturz und Verletzungen kommen.
Achten Sie beim Einsetzen der Batterien darauf, dass die Pole + (positiv) und - (negativ) richtig ausgerichtet sind.
Wird die Batterie falsch eingesetzt, kann es zu einer Überhitzung kommen, Batterieflüssigkeit kann auslaufen oderdie Batterien können beschädigt werden, was zu Schäden an der Waage oder Verletzungen führen kann.
Diese Waage wurde für den persönlichen Gebrauch entwickelt.
Lagern Sie die Waage außerhalb der Reichweite kleiner Kinder.
Lagern Sie die Waage nicht vertikal. Wenn die Waage aus irgendeinem Grund vertikal gelagert wird, achten Sie
darauf, dass sie nicht umfällt.
Legen Sie bei der Lagerung keine Gegenstände auf der Waage ab.
Reinigen Sie die Waage vor der Verwendung, wenn Sie die Waage gemeinsam mit einer Person verwenden, dieeine Infektionskrankheit, wie z. B. eine Hautkrankheit, an den Füßen usw. hat.
(Ausführliche Informationen zur Reinigung der Waage finden Sie unter „Wartung der Waage (S. 13)“.)
Um Ihr Körpergewicht genau zu messen, verwenden Sie die Waage auf einem ebenen, harten Untergrund.
Verwenden Sie sie nicht auf einem Teppich oder anderen weichen Untergrund.

Deutsch 2

Funktion zur drahtlosen
Kommunikation
Diese Waage verwendet die drahtlose Bluetooth-Technologie zur Kommunikation mit einem Empfänger.
Am Ende der Messung werden die Messdaten automatisch von der Waage an den Empfänger gesendet.
Bluetooth ist eine Kommunikationstechnologie, durch die Geräte drahtlos kommunizieren können.
Ausführliche Informationen zu Bluetooth finden Sie unter „Kommunikation mit der Waage (S. 11)“.

Empfänger
Hierbei handelt es sich um ein Gerät wie einen Computer oder ein Smartphone, auf dem eine Anwendung
zur Aktivierung der Bluetooth-Kommunikation mit der Waage heruntergeladen und installiert wurde.

Vorsichtshinweise zur Drahtlosfunktion
Warnung
Verwenden Sie die Waage nicht an Orten, an denen eine drahtlose Kommunikation verboten ist, wie z. B. in
Flugzeugen oder in Krankenhäusern.
Die Waage kann negative Auswirkungen auf elektronische Geräte oder medizinische elektrische Geräte haben.

AchtungWenn die Kommunikationsfunktion dieser Waage in der Nähe eines drahtlosen Geräts verwendet wird , das im
2,4-GHz-Bereich kommuniziert, kann die Kommunikationsgeschwindigkeit beider Geräte sinken. Verwenden Sie die
Waage nicht in der Nähe eines Mikrowellengeräts, bei dem es zu einer Störung durch ein Magnetfeld, statische
Elektrizität oder Funkwellen kommen kann. (Das Signal erreicht den Empfänger möglicherweise aufgrund derUmweltbedingungen nicht.)※ A&D übernimmt keine Verantwortung für Verluste, die durch Bedienungsfehlfunktionen oder
Datenverlust aufgrund der Verwendung dieser Waage entstanden sind.
※ Es wird nicht garantiert, dass sich diese Waage mit allen Bluetooth-kompatiblen Geräten verbinden kann.

Zubehör
Wenn Zubehörteile fehlen oder beschädigt sind, wenden Sie sich an den Produktanbieter oder an das
Vertriebsteam von A&D.
Precision Health Scale

Instruction Manual
Mode d'emploi

Manuale d’uso

Italiano

Bedienungsanleitung

DeutschManual instructivo

Español Français English

UC-351PBT-Ci

1WMPD4002776

Bedienungsanleitung

Alkalibatterien x 4

Deutsch 3

Bezeichnung der Teile
Waage
Batteriefach
GriffEinstellungsschalter

Füße

Anzeige
SymbolBluetooth-Kommun
ikationSymbol Stabil
Batterieanzeige

EinheitAnzeigesymbole
Anzeige
Symbol○Name

Beschreibung
Dieses Symbol wird angezeigt, wenn die
Messanzeige stabil ist.
Die normale Batterieleistungsanzeige mit Ladung
wird angezeigt.

Symbol Bereit/Stabil
Batterieanzeige↓
Anzeige niedriger BatteriestandWenn dieses Symbol angezeigt wird, tauschen Sie
alle vier Batterien gegen neue aus.

BlinktSymbol Bluetooth-Kommunikation


Unterseite der Waage
SymbolName

Beschreibung

WEEE-Kennzeichnung-----

Deutsch 4

Einsetzen (oder Austauschen) der
Batterien
Sie die Batteriefachabdeckung an
1. Öffnender Unterseite der Waage.

Sie vier neue Alkalibatterien der
2. Setzen
Größe AA ein, sodass die positiven (+) und

negativen (-) Pole entsprechend den
Symbolen im Batteriefach ausgerichtet sind.

Sie die Batteriefachabdeckung.
3. Schließen
Die Waage wechselt zum Kopplungsmodus, wenn die
Batterien eingelegt wurden.
Um die Waage mit dem Empfänger zu koppeln, schlagen Sie
unter „Registrieren bei einem Empfänger (Kopplung)(S. 6)“ nach. Wenn Sie die Waage nicht mit einem Empfänger
koppeln möchten, schlagen Sie unter „Überspringen derRegistrierung bei einem Empfänger (Kopplung) (S. 7)“ nach.

AchtungSetzen Sie jede Batterie so ein, dass die positiven (+) und negativen (-) Pole entsprechend den
Symbolen im Batteriefach ausgerichtet sind.
Eine unsachgemäße Installation der Batterien kann zum Auslaufen der Batterien führen.
Tauschen Sie alle vier Batterien gegen neu aus, wenn das Symbol
angezeigt wird.
Verwenden Sie nicht neue und alte Batterien zusammen. Andernfalls kann es zu Schäden an der
Waage kommen oder die Nutzungsdauer der Batterien kann sich verkürzen.
Achten Sie darauf, dass Sie Alkali-Mangan-Batterien der Größe AA verwenden. Verwenden Sie
keine anderen Batteriearten.
Verwenden Sie niemals wiederaufladbare Batterien (Akkus) in der Waage.
Die Nutzungsdauer der Batterien hängt von der Umgebungstemperatur, der Batterieart und der Verwendung ab.
Die Batterien im Lieferumfang des Geräts dienen zum Test des Betriebs der Waage und können
eine eingeschränkte Nutzungsdauer aufweisen.
Entnehmen Sie die Batterien aus dem Batteriefach, wenn die Waage längere Zeit nicht verwendet
wird. Wenn die Waage mit eingelegten Batterien gelagert wird, kann dies dazu führen, dass die
Batterieflüssigkeit aus den Batterien ausläuft, was zu Schäden an der Waage führen kann.

Automatische Abschaltfunktion
Die Waage verfügt über eine „Automatische Abschaltfunktion“.
Nachdem die Messergebnisse 15 Sekunden lang angezeigt wurden, wird die Waage automatischausgeschaltet.
Deutsch 5

Kopplung mit einem Empfänger
Diese Waage muss mit einem Empfänger gekoppelt werden, um die Bluetooth-Kommunikation mit dem
Empfänger nutzen zu können. Die Verbindungder Waage mit einem Empfänger wird als Kopplung
bezeichnet.
Wenn die Kopplung abgeschlossen ist, werden die Messdaten von der Waage bei jeder Messung
automatisch an das gekoppelte Gerät gesendet, wenn der Empfänger einen Status aufweist, in dem erÜbertragungen empfangen kann.
Schlagen Sie in der Anleitung des Empfängers nach und stellen Sie ihn so ein, dass die Kopplungaktiviert ist. Wenn der Empfänger über einen „Kopplungsassistenten“ verfügt, verwenden Sie denAssistenten, um die Kopplung abzuschließen.
Positionieren Sie den Empfänger und die Waage so nah wie möglich beieinander, um die Kopplung
durchzuführen.

Sie vier Alkali-Mangan-Batterien der
1. LegenGröße AA in die Waage ein.
Ausführliche Informationen finden Sie unter „Einsetzen (oderAustauschen) der Batterien (S. 5)“.

Sie, dass „Pr“ angezeigt wird.
2. Bestätigen„ “ blinkt.

≪♪Piepton≫

Die Waage kann vom Empfänger erkannt werden.
Die Kopplung kann anhand der folgenden Schritteübersprungen werden.
i) Drücken Sie den Einstellungsschalter , wenn
„Pr“ angezeigt wird.

3. Starten Sie die Kopplung beim Empfänger.

Befolgen Sie die Kopplungsanweisungen für den Empfänger, um
nach der Waage zu suchen, sie auszuwählen und die Kopplung
durchzuführen.
Wenn der Empfänger einen PIN-Code verlangt, geben Sie„39121440“ ein.

Sie, dass „End“ angezeigt wird.
4. Bestätigen
Die Kopplung ist abgeschlossen.

≪♪Piepton Piepton≫

Wenn „E-10“ oder „E-11“ angezeigt wird, führen Sie den
Kopplungsvorgang erneut ab Schritt 1 durch.

※ Diese Waage kann mit zwei Empfängern gekoppelt werden.
※ Wenn die Waage mit einem dritten Empfänger gekoppelt wird, bleiben die Informationen desEmpfängers, mit dem zuletzt eine Kommunikation erfolgt ist, gespeichert.
※ Wenn die Waage mit dem Empfänger gekoppelt wurde, muss keine erneute Kopplung ausgeführt
werden, wenn die Batterien ausgetauscht wurden.

Deutsch 6

Vorsichtshinweise zur Kopplung
Schalten Sie andere Bluetooth-Geräte aus, wenn Sie eine Kopplung durchführen.
Es können nicht mehrere Geräte gleichzeitig gekoppelt werden.
Wenn die Messdaten nicht empfangen werden können, führen Sie die Kopplung erneut durch.
Nachdem die Kopplung abgeschlossen ist, erfolgt die gesamte Kommunikation automatisch.
Um die Kopplung eines Empfängers aufzuheben, schlagen Sie unter „Aufheben der Kopplung(S. 10)“ nach.
Falls es zu dem unwahrscheinlichen Fall kommt, dass die Waage Funkwellenstörungen für andere
drahtlose Gerät verursacht, sollten Sie entweder den Standort der Waage ändern oder die
Verwendung der Waage sofort beenden.

Überspringen der Kopplung mit
einem Empfänger
Sie vier Alkali-Mangan-Batterien der
1. LegenGröße AA in die Waage ein.
Ausführliche Informationen finden Sie unter „Einsetzen (oderAustauschen) der Batterien (S. 5)“.

Sie den Einstellungsschalter, wenn
2. Drücken
„Pr“ angezeigt wird.
Drücken Sie den Einstellungsschalter, um das Gerät
auszuschalten.

Deutsch 7≪♪Piepton≫

Einstellen des Geräts
Sie, dass die
1. Überprüfen
Waage ausgeschaltet ist,
und halten Sie den
Einstellungsschaltergedrückt.

≪♪Piepton≫

Wenn der Einstellungsschalter
fortlaufend gedrückt wird, werden„unit“ und „uPr“ abwechselndangezeigt.

Sie den
2. Lassen
Einstellungsschalter los,≪♪Piepton≫

wenn „unit“ angezeigt
wird.
Die Anzeige wechselt von „unit“ zuder Einheit, die derzeit eingestellt
ist.

kg

Sie den
3. Drücken
Einstellungsschalter, um

kg

die Einheit auszuwählen,die eingestellt werden soll.
Die Einheit wechselt mit jeder
Betätigung desEinstellungsschalters.
Warten Sie 3 Sekunden, nachdem
Sie die Einheit eingestellt haben.

lbSie, dass
4. Bestätigen„End“ angezeigt wird.

≪♪Piepton
Piepton≫

Die Einstellungen wurden
gespeichert.
Nach drei Sekunden ohne
Bedienung bestätigt die Waage dieausgewählte Einheit mit zwei
Pieptönen, zeigt „End“ an undschaltet sich automatisch aus.
Deutsch 8

Messen des Gewichts
Auf die Waage stellenSie, dass die
1. Überprüfen
Waage ausgeschaltet ist,

und stellen Sie sichvorsichtig auf die Waage.
Die Messung beginnt, wenn Sie auf
die Waage steigen.
Ein Fortschrittsbalken wird während≪♪Piepton≫

Sie von der Waage
2. Steigen
herunter, wenn „StEP“ und≪♪ Piepton
Piepton
Piepton≫der Messung angezeigt. Stehen Sie
während der Messung gerade und so
still wie möglich.

„oFF“ abwechselndangezeigt werden und drei
Pieptöne zu hören sind.
Wenn Sie nicht von der Waage
heruntersteigen, piept die Waage
alle zwei Sekunden.
Die Waage schaltet sichautomatisch nach 2 Minuten aus.
Alle Anzeigesegmente werdenangezeigt, wenn Sie von der
Waage heruntersteigen.
Berühren Sie die Waage nicht,
während alle Anzeigesegmenteangezeigt werden.
Andernfalls kann es zu einer
Fehlmessung kommen.

in
c
lb
kg

Sie das3. Überprüfen
Messergebnis.

kgDie Waage schaltet sich automatisch15 Sekunden nach der Anzeige desMessergebnisses ab.
Die Daten werden automatisch an den
Empfänger gesendet, wenn derEmpfänger in Empfangsbereitschaft ist.
Wenn die Waage mit keinem
Empfänger gekoppelt ist, werden die
Daten im Speicher der Waage
gespeichert.
(Siehe „Übertragen vorübergehend
gespeicherter Daten (S. 10)“.)
Die im Speicher der Waage
gespeicherten Daten können nicht auf
der Waage angezeigt werden.
Sie können das Gerät auchausschalten, indem Sie den
Einstellungsschalter drücken.
Deutsch 9

Sonstige Funktionen
Aufheben der KopplungSie, dass die
1. Überprüfen
Waage ausgeschaltet ist,
und halten Sie den
Einstellungsschaltergedrückt.

≪♪Piepton≫

Wenn der Einstellungsschalter
fortlaufend gedrückt wird, werden„unit“ und „uPr“ abwechselndangezeigt.

Sie den
2. Lassen
Einstellungsschalter los,≪♪Piepton≫

Sie, dass
3. Bestätigen„End“ angezeigt wird.

≪♪Piepton Piepton≫

wenn „uPr“ angezeigt
wird.

Die Kopplung mit dem Empfänger
wurde aufgehoben.
Wenn „E-10“ oder „E-11“ angezeigt
wird, führen Sie den Vorgang zum
Aufheben der Kopplung erneut ab
Schritt 1 aus.
Die Waage schaltet sichautomatisch aus.

Übertragen vorübergehend gespeicherter Daten
Wenn die Messdaten nicht an einen Empfänger gesendet werden können, kann die Waage die Daten
vorübergehend für bis zu 200 Messungen im integrierten Speicher speichern. Wenn die Waage Daten an
einen Empfänger übertragen kann, werden alle Daten übertragen. Wenn die Menge der vorübergehend
gespeicherten Daten 200 Messungen übersteigt, werden die ältesten Daten gelöscht und durch neue
Daten ersetzt.
※ Die gespeicherten Daten können mit der Waage nicht angezeigt werden.
※ Der maximale Umfang der vorübergehend speicherbaren Daten kann abhängig vom Empfänger
variieren.
Deutsch 10

Kommunikation mit der Waage
Diese Waage ist mit einer von Continua zertifizierten, drahtlosen Bluetooth-Kommunikationsfunktion
ausgestattet. Die Messdaten können durch die Verbindung mit einem Bluetooth-Gerät übertragenwerden, dass mit den Continua-Verbindungsspezifikationen kompatibel ist.
Geräte, die von Continua zertifiziert sind, tragen das Continua-Logo.
Bluetooth-Geräte tragen das Bluetooth-Logo.
Ausführliche Informationen zur Kopplung der Waage mit einem Empfänger, der mit Bluetoothausgestattet ist, finden Sie unter „Kopplung mit einem Empfänger (S. 6)“.
Schlagen Sie auch in der Kopplungsanleitung des Empfängers nach, mit dem Sie eine Kopplung
durchführen möchten.

Sicherstellen einer guten VerbindungStellen Sie sicher, dass sich im Bereich zwischen der Waage und dem Empfänger keine Hindernisse
befinden. Die Kommunikationsentfernung kann sich aufgrund der Gebäudebauweise oder anderer
Hindernisse verringern. Insbesondere bei verstärkten Betonwänden usw. ist eine Kommunikation
unter Umständen nicht möglich.
Verwenden Sie die Bluetooth-Kommunikation nicht in der Nähe eines WLAN- oder eines anderendrahtlosen Geräts, in der Nähe von Geräten, die elektromagnetische Wellen aussenden, wie z. B.
Mikrowellengeräte, an Orten mit sehr vielen Hindernissen oder mit schlechten Empfangsbedingungen
für Funkwellen. Andernfalls kann die Verbindung häufig unterbrochen werden, die
Kommunikationsgeschwindigkeit kann deutlich absinken und es kann zu Fehlern kommen.
Wenn die Geräte in der Nähe eines IEEE802.11g/b/n-WLAN-Geräts verwendet werden, verursachen
die Geräte gegenseitig Funkwellenstörungen, die Kommunikationsgeschwindigkeit kann absinken
oder die Verbindung ist möglicherweise nicht möglich. Schalten Sie in dieser Situation das nichtverwendete Gerät aus oder wechseln Sie den Standort, an dem das Gerät verwendet wird.
Wenn die Waage nicht ordnungsgemäß kommuniziert, wenn sie in der Nähe eines drahtlosen Geräts
oder einer Sendestation verwendet wird, wechseln Sie den Standort der Waage.

Achtung
Die Kommunikationsentfernung zwischen der Waage und dem Empfänger hängt von derBluetooth-Ausgangsklasse des Empfängers ab.
Für Bluetooth-Empfänger der Klasse 1: Max. 100 m
Für Bluetooth-Empfänger der Klasse 2: Max. 10 m
Diese Entfernung kann sich aufgrund der Umgebung verringern; prüfen Sie daher, ob die
Entfernung ausreichend gering ist, um Daten nach der Messung zu übertragen.

Zeiteinstellungen
Die Waage verfügt über eine integrierte Uhr. Das Datum und die Uhrzeit der Messung sind in den
Messdaten enthalten. Die Uhr wird mit der Uhr des Empfängers synchronisiert. Ausführliche
Informationen zu den technischen Daten der Uhr finden Sie in der Bedienungsanleitung desEmpfängers.

Deutsch 11

Technische Daten
Modell

UC-351PBT-Ci

Maximale Kapazität

150 kg / 330 lb

Minimale Anzeige

0,1-kg-Schritte von 0 bis 100,0 kg
0,2-kg-Schritte von 100,0 bis 150,0 kg
0,2-kg-Schritte von 0 bis 220,0 lb
0,5-lb-Schritte von 220,0 bis 330,0 lb

Drahtlose
Kommunikation

WML-C40AH (MITSUMI ELECTRIC CO., LTD.)
Bluetooth Version 2.1 Klasse 1
Kompatibel mit SSP HDP
Frequenzbereich: 2402 MHz bis 2480 MHz
Maximale HF-Ausgangsleistung: 20 dBm

Messeinheiten

kg oder lb

Stromversorgung

Vier Alkali-Mangan-Batterien der Größe AA

Nutzungsdauer
BatterieCa. 1 Jahr (zwei Messungen pro Tag, wenn Alkali-Mangan-Batterienverwendet werden)

Abmessungen350 (B) ×350 (T) ×39 (H) mm

Gewicht

Ca. 2,3 kg (einschließlich Batterien)

Zubehör

Bedienungsanleitung, Alkali-Mangan-Batterien
Einhaltung europäischer Richtlinien
Die Waage entspricht der gesetzlichen EMV-Richtlinie (Elektromagnetische Verträglichkeit)
2014/30/EU.
Hiermit erklärt MITSUMI ELECTRIC CO., LTD., dass der Funkanlagentyp WML-C40AH derRichtlinie 2014/53/EU entspricht. Der vollständige Text der EU-Konformitätserklärung ist
unter der folgenden Internetadresse verfügbar:
http://www.aandd.jp/products/manual/manual_medical.html
Hinweise: Nur zur Verwendung in Gebäuden. Unter extremen elektromagnetischen Einflüssen
kann der angezeigte Wert negativ beeinflusst werden.

Einhaltung der FCC-Richtlinien
Die Waage entspricht Teil 15 der FCC-Richtlinien und umfasst FCC ID POOWML-C40.

Einhaltung Industry Canada
IC: 4250A-WMLC40.

Deutsch 12

Bevor Sie sich an den
Kundendienst wenden
Symptome

Maßnahme

E wird angezeigt

Die maximale Kapazität wurde überstiegen (150 kg/330 lb).

E-0 wird angezeigt

Die maximale Kapazität wurde überstiegen (150 kg/330 lb).

Err wird angezeigt

Die maximale Kapazität wurde überstiegen (150 kg/330 lb).

E-10 wird angezeigt
E-11 wird angezeigt

Die Kopplung oder Aufhebung der Kopplung wurde nicht
ordnungsgemäß abgeschlossen. Entnehmen Sie die Batterien undsetzen Sie sie wieder ein.

H wird angezeigt

Die maximale Kapazität wurde überstiegen (150 kg/330 lb).

wird angezeigt
Wenn derEinstellungsschalter
betätigt wird, wird nichts
angezeigt
Die Messergebnisse
weichen stark
von den erwarteten
Werten ab.

Entnehmen Sie die Batterien und setzen Sie sie wieder ein. Odertauschen Sie alle vier Batterien gegen neue Alkali-Mangan-Batterien der
Größe AA aus.

Stellen Sie die Waage auf eine feste und ebene Fläche.
Entfernen Sie alle Fremdkörper, die möglicherweise die Unterseite der
Waage berühren.

Die Messung beginnt nicht, Haben Sie den Einstellungsschalter betätigt, als Sie auf die Waage
wenn ich auf die Waage
gestiegen sind?
steige.
Stellen Sie die Waage auf einen harten Boden und drücken Sie vor der
Messung den Einstellungsschalter.

Wartung der Waage
Wenn die Waage schmutzig wird, befeuchten Sie ein Tuch mit Wasser oder einem milden
Reinigungsmittel, wringen Sie es sorgfältig aus und wischen Sie die Waage dann sauber.
Trocknen Sie die Waage mit einem weichen, trockenen Tuch ab, nachdem Sie den Schmutz mit dem
feuchten Tuch entfernt haben.
Verwenden Sie niemals Alkohol, Benzol, Verdünner oder scharfe Chemikalien, um die Waage zu
reinigen.
Entnehmen Sie die Batterien aus dem Batteriefach, wenn die Waage längere Zeit nicht verwendet
wird.

Deutsch 13

Deutsch 14

MEMO

Deutsch 15

1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119

OX14

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409
OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd

爱安德技研贸易（上海）有限公司

中国 上海市浦东新区浦东大道 138 号永华大厦 21 楼 A 室 邮编 200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, Chin
电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599