Digital Blood Pressure Monitor
Model UA-767JP
Instruction Manual
Original

使用手冊

English

中文

翻譯

사용설명서
번역본

‫ﺩﻟﻳﻝ ﺍﻟﺗﻌﻠﻳﻣﺎﺕ‬‫ﺗﺭﺟﻣﺔ‬1WMPD4002956B

한국어‫ﻋﺭﺑﻲ‬

Contents
Dear Customers .................................................................... 2
Preliminary Remarks ............................................................. 2
Precautions ........................................................................... 2
Parts Identification ................................................................ 5
Symbols ................................................................................ 6
Operation Mode .................................................................... 8
Using the Monitor ............................................................... 10
Installing / Changing the Batteries ................................. 10
Connecting the Air Hose ................................................ 11
Connecting the AC Adapter ............................................ 11
Adjusting the Built-in Clock ........................................... 12
Selecting the Correct Cuff Size ....................................... 13
Applying the Arm Cuff ................................................... 13
How to Take Accurate Measurements ............................. 15
Measurement ................................................................. 15
After Measurement ........................................................ 15
Measurements .................................................................... 16
Normal Measurement ..................................................... 16
Measurement with the Desired Systolic Pressure ............ 17
Notes for Accurate Measurement ................................... 17
Recalling the Memory Data ................................................. 18
What Is The IHB/AFib Indicator? .......................................... 20
What Is The AFib? ............................................................... 20％IHB/AFib .......................................................................... 20
Pressure Bar Indicator ......................................................... 21
WHO Classification Indicator ............................................... 21
About Blood Pressure.......................................................... 22
What is Blood Pressure? ................................................. 22
What is Hypertension and How is it Controlled? .............. 22
Why Measure Blood Pressure at Home?........................... 22
WHO Blood Pressure Classification ................................. 22
Blood Pressure Variations............................................... 22
Troubleshooting ................................................................. 23
Maintenance ....................................................................... 24
Technical Data .................................................................... 24

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure
monitor, one of the most advanced monitors available today. Designed
for ease of use and accuracy, this monitor will facilitate your daily bloodpressure regimen.
We recommend that you read through this manual carefully before
using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity. (0123:
The reference number to the involved notified body)
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in thehome healthcare environment.
This device is designed to measure blood pressure and pulse rate ofpeople for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremesin temperature, humidity, direct sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff or storing the hose tightly twisted for long
periods, as such treatment may shorten the life of the components.
Take care to avoid accidental strangulation of babies or infants with the hose.
Do not twist the air hose during measurement. This may cause injury due
to continuous cuff pressure.
The device and cuff are not water resistant. Prevent rain, sweat and waterfrom soiling the device and cuff.
Do not replace the batteries during use of the device.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
Wireless communication devices, such as home networking devices,mobile phones, cordless phones and their base stations, walkie-talkies
can affect this blood pressure monitor. Therefore, a minimum distance of
30 cm should be kept from such devices.
English 2

When reusing the device, confirm that the device is clean.
Used equipment, parts and batteries are not treated as ordinary
household waste, and must be disposed of according to the applicable
local regulations.
When the AC adapter is used, make sure that the AC adapter can bereadily removed from the electrical outlet when necessary.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the arm must be squeezed by the cuff hardenough to temporarily stop blood flow through the artery. This may cause
pain, numbness or a temporary red mark to the arm. This condition willappear especially when measurement is repeated successively. Any pain,
numbness, or red marks will disappear with time.
Measuring blood pressure too frequently may cause harm due to blood flowinterference. Check that the operation of the device does not result in
prolonged impairment of blood circulation, when using the device repeatedly.
If you have had a mastectomy, please consult a doctor before using thedevice.
Do not let children use the device by themselves and do not use the device
in a place within the reach of infants. It may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
Unplug the AC adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially causeburns.
Allow the device to adapt to the surrounding environment before use (about
one hour).
Clinical testing has not been conducted on newborn infants and pregnant
woman. Do not use on newborn infants or pregnant woman.
Do not touch the batteries, the DC jack, and the patient at the same time.
That may result in electrical shock.
Do not inflate without wrapping the cuff around the upper arm.

Contraindications

The following are precautions for proper use of the device.
Do not apply the cuff on an arm with another medical electrical equipment
attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult adoctor before using the device, to avoid medical problems.
Do not self-diagnose the measurement results and start treatment by yourself.
Always consult your doctor for evaluation of the results and treatment.
Do not apply the cuff on an arm with an unhealed wound.

English 3

Do not apply the cuff on an arm receiving an intravenous drip or bloodtransfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, suchas a high-pressure oxygen chamber or an oxygen tent. It may cause a fireor explosion.

English 4

Parts Identification
ButtonButtonSET Button

AC Adapter Plug
DC Jack

Air Connector Plug

START Button

Air Socket
Display
Air Hose

Proper Fit Range

Artery Position Mark
Index MarkArm Cuff
Battery Compartment

Battery Cover

1.5V Batteries(R6P, LR6 or AA)

Display

Average
MEMORYIHB/AFib symbol
Systolic
Pressure%IHB/AFib
Cuff Fit Error Symbol

Diastolic
Pressure

Movement Error Symbol
Heart MarkWHO Classification Indicatorand Pressure Bar Indicator

Pulse Rate
Date &
Clock Display

Battery Indicator

User
English 5

Symbols
Symbols that are printed on the device caseSymbols
SYS
DIA
PUL

IP

Function / MeaningStandby and Turn the device on.
Systolic blood pressure in mmHg
Diastolic blood pressure in mmHg
Pulse per minuteBattery installation guide
Direct current
Type BF: Device, cuff and tubing are designed to provide
special protection against electrical shocks.
EC directive medical device label
EU-representative
Manufacturer
Date of manufactureInternational protection symbol
WEEE label

SN

Serial number
Refer to instruction manual/booklet
Polarity of DC jack
Keep drySymbols that appear on the display
Symbols

Function / Meaning
Recommended Action
Measurement is in
Appears while measurement is inprogress. It blinks when the pulse is progress. Remain as
still as possible.
detected.
IHB/AFib symbol
Appears when an irregular heartbeat
is detected.
It may light when a very slight vibration
like shivering or shaking is detected.
The reading may yield
an incorrect value. Take
Appears when a body or arm
another measurement.
movement is detected.
Remain still during
measurement.
English 6

Symbols
Symbols

Function / Meaning
Appears during measurement when
the cuff is attached loosely.

Recommended Action
The reading may yield anincorrect value. Apply the
cuff correctly, and takeanother measurement.

Detected rate of IHB/AFib in memory%IHB/AFib =Number of detected
IHB/AFibs in memory

x100 [%]

Total number

User
Previous measurements stored in memory.
Average data
FULL BATTERY
The battery power indicator during
measurement.
LOW BATTERY
The battery is low when it blinks.
Unstable blood pressure due tomovement during measurement.

Replace all batteries with newones when the mark blinks.
Take another measurement.
Remain very still during
measurement.

The systolic and diastolic values arewithin 10 mmHg of each other.
The pressure value did not increase
Apply the cuff correctly,
during the inflation.
and take anothermeasurement.
The cuff is not applied correctly.
PUL DISPLAY ERROR
The pulse is not detected correctly.

AM

Remove the batteries andpress the START button,
Blood pressure monitor internal error and then install the batteries
again. If the error still
appears, contact the dealer.
Data taken between 4:00 and 9:59

PM

Data taken between 18:00 and 1:59
English 7

Operation Mode
1. Normal Measurement
Press the START button. Blood pressure is measured and the data isstored in memory. This device can store the last 60 measurements foreach of the four users in memory.

2. Recalling the Data
Press theor button to recall the data inmemory. The average of all measurements is
displayed, as indicated in the figure at the right.
Then, each time thebutton is pressed, thememory data is displayed as follows.
Most recent data (No.n, in the example, No.35)

Last data (No.1)
For details on recalling the data, refer to“Recalling the Memory Data”.

3. Changing User for Memory Display
Press the SET button while a memory is displayed.
The user is changed, and the average ofmeasurement values for that user is displayed.

English 8

Operation Mode
4. Deleting Data Stored in Memory
In standby mode, press both theand buttons.
Themark, battery indicator and user mark will appear.
When you would like to delete the memory data of thecurrently displayed user, press and hold both theandbuttons until the illuminatedmark starts blinking.

5. Measurement with the Desired Systolic Pressure
Refer to page 17 for measurement with the desired systolic pressure.

6. User Selection
Press the SET button before measuring blood pressure to select a userfrom 1 to 4.
In standby mode, press the SET button.

Each time the SET button is pressed, the user changes.
Press the START button to set to the currently
displayed user.

English 9

Using the Monitor
Installing / Changing the Batteries
1. Remove the battery cover.
2. Remove the used batteries andinsert new batteries into the battery
compartment as shown, taking
care that the polarities (+ and -) arecorrect.
Use only R6P, LR6 or AA batteries.

Step 1
Step 3

Step 1
Step 2

3. Attach the battery cover.
CAUTION
Insert the batteries as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(LOW BATTERY mark) blinks on the display, replace all batteries
with new ones. Do not mix old and new batteries. It may shorten the battery
life, or cause the device to malfunction.
Replace the batteries two seconds or more after the device turns off.
If
(LOW BATTERY mark) appears even after the batteries are replaced,make a blood pressure measurement. The device may then recognize thenew batteries.
(LOW BATTERY mark) does not appear when the batteries are drained.
The battery life varies with the ambient temperature and may be shorter at
low temperatures. Generally, four new LR6 batteries will last approximately
for one year when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device arefor testing monitor performance and may have a limited life.
Remove the batteries if the device is not to be used for a long time.
The batteries may leak and cause a malfunction.

English 10

Using the Monitor
Connecting the Air Hose

Air socket

Insert the air connector plug into
the air socket firmly.

Air connector plug

Connecting the AC Adapter
Insert the AC adapter plug into the DC
jack.
Next, connect the AC adapter to anelectrical outlet.

DC jack
AC adapter plug

Use the specified AC adapter. (Refer to page 25.)
When disconnecting the AC adapter from the electrical outlet, grasp
and pull the AC adapter body out of the outlet.
When disconnecting the AC adapter plug from the blood pressure
monitor, grasp and pull the AC adapter plug out of the monitor.

English 11

Using the Monitor
Adjusting the Built-in Clock
Adjust the clock prior to use.

Press and hold

1. Press and hold the SET button until the year
starts blinking.
or button.
2. Select the year using thePress the SET button to set the current year
and move to month/day selection. The datecan be set anywhere between the years
2013 and 2059.
3. Select the month using theor button.
Press the SET button to set the current
month and move to day selection.
4. Select the day using theor button.
Press the SET button to set the current day
and move to hour/minute selection.
or button.
5. Select the hour using thePress the SET button to set the current hourand move to minute selection.

or

or

Month

or

or

Day

Hour

or

6. Select the minute using theor button.
Press the START or SET button to turn thedevice off.
Holding down theorbutton willchange the value continuously.
Note: After three minutes of non-operation,
the device will turn off automatically.
When the clock has not beenset, the clock display indicatesdashes as shown to the right.

START

or

Turn off
Pressing the START
button will turn thedevice off anytime.

When using the device for the first time, the clock is not adjusted.
When the device is disconnected from the power supply for more than 30
seconds, the set date and time will be erased.
When the set date and time is erased, please adjust again.

English 12

Using the Monitor
Selecting the Correct Cuff Size
Using the correct cuff size is important for an accurate reading. If the cuff is
not the proper size, the reading may yield an incorrect blood pressure value.
The arm size is printed on each cuff.
The index
and proper fit range, on the cuff, tell you if you are applying thecorrect cuff. (Refer to "Symbols that are printed on the cuff" on the next page)
If the index
points outside of the range, contact your local dealer topurchase a replacement cuff.
The arm cuff is a consumable. If it becomes worn, purchase a new one.
Arm Size
Recommended Cuff Size
Catalog Number
31 cm to 45 cm
Large Adult Cuff
CUF-F-LA
22 cm to 42 cm
Wide Range Cuff
CUF-I
22 cm to 32 cm
Adult Cuff
CUF-F-A
Arm size: The circumference at the biceps.
Note:
Model UA-767JP is not designed for using a small cuff.

Applying the Arm Cuff
1. Wrap the cuff around the upper arm,
about 1-2 cm above the inside of theelbow, as shown.
Place the cuff directly against theskin, as clothing may cause a faintpulse, and result in a measurement
error.

Do not roll up shirtsleeve tightly.
Cuff

Air hose
Artery position mark2. Constriction of the upper arm,
caused by tightly rolling up a
shirtsleeve, may prevent accurate
readings.
3. Confirm that the index
pointswithin the proper fit range.

English 13

1-2 cmSecure the fabricfastener firmly.

Index

and proper fit range

Using the Monitor
Symbols that are printed on the cuff
Symbols
Function / Meaning

Recommended Action
mark on the
Set theartery of the upper arm orin line with the ring finger
on the inside of the arm.

Artery position mark
▲
REF

Index
Catalog number
Proper fit range for the adult cuff.
It's printed on the adult cuff.
Over range printed on the adult cuff
and wide range cuff.

Use the large adult cuff
instead of the adult cuff orwide range cuff.

Proper fit range for the wide range cuff.
It's printed on the wide range cuff.
Proper fit range for the large adult cuff.
It's printed on the large adult cuff.
Under range printed on the adult cuff
and wide range cuff.
Under range printed on the large adult Use the adult cuff instead
cuff.
of the large adult cuff.
Lot number
Large adult cuff

A

Proper fit range

L

Wide range cuff
Adult cuff

S

W

L

S

A

L

English 14

Using the Monitor
How to Take Accurate Measurements
For the most accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross yourlegs. Keep your feet on the floor and straighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same height as your heart.
Remain still and keep quiet during measurement.
Do not measure immediately after physical exercise or a bath. Rest fortwenty or thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

Measurement
During measurement, it is normal for the cuff to feel very tight. (Do not be
alarmed).

After Measurement
After measurement, press the START button to turn the device off. Afterone minute of non-operation, the device will turn off automatically.
Remove the cuff and record your data.
Note: Allow at least three minutes between measurements on the same person.

English 15

Measurements
Before measurement, read “Notes for Accurate Measurement” on the next page.

Normal Measurement
1.

Press the SET button to select a user from 1 to 4.

At heart level
Press START

2.

Place the cuff on the arm (preferably the left arm).
Sit quietly during measurement.
All of the display
3. Press the START button.
segments displayed
All of the display segments are displayed.
Zero (0) is displayed blinking briefly.
The display changes, as indicated in the
figure at the right, as the measurement
Zero display
begins. The cuff starts to inflate. It is
Starts inflation
normal for the cuff to feel very tight. A
pressure bar indicator is displayed, on theleft edge of the display, during the inflation.
Note: If you wish to stop inflation at any time,press the START button again.
Pressurizing
4. When inflation is complete, deflation starts(heart mark) blinks,
automatically andindicating that the measurement is inprogress. Once the pulse is detected, themark blinks with each pulse beat.
Note: If an appropriate pressure is not obtained, thedevice starts to inflate again automatically.
To avoid re-inflation, see “Measurement with
the Desired Systolic Pressure” on the next
page.
Measurement
5. When the measurement is complete,in progress
the systolic and diastolic pressure
readings and pulse rate are displayed.
The cuff exhausts the remaining air andSystolic pressure
deflates completely.
Diastolic pressure
6. If you do not want to store the measurement
WHO classificationorbutton.
data in memory, press thePulse rate
7. Press the START button to turn the device off.
After one minute of non-operation, theExhausts remaining
device will turn off automatically.
air automaticallyNote: Allow at least three minutes between measurements on the same person.
English 16

Measurements
Measurement with the Desired Systolic Pressure
Model UA-767JP is designed to detect the pulse and to inflate the cuff toa systolic pressure level automatically.
Use this method when re-inflation occurs repeatedly or when the results
are not displayed even if the pressure decreases to 20 mmHg or less.
1.

Place the cuff on the arm at heart level(preferably the left arm).

At heart level2.

Press and hold the START button until a
number about 30 to 40 mmHg higher thanyour expected systolic pressure appears.

Press and holdthe button

3.

When the desired number is reached,release the START button to startmeasurement. Continue to measure
your blood pressure as described on theprevious page.

Notes for Accurate Measurement

START

Release the button
at the desired
systolic pressure

Refer to the previous pagefor measurement

Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five to ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.
Should the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. Refer to page 7 for thedescription of symbols.
This blood pressure monitor is intended for use by adults. Consult withyour physician before using this device on a child. A child should not use
this device unattended.
The automatic blood pressure monitor’s performance may be affected byexcessive temperature or humidity, or altitude.
English 17

Recalling the Memory Data
Note: This device stores the last 60 measurements for each of the four users
in memory.
or button.
1. Press thePress orThe average of all measurements andthe number of data are displayed.
Average systolic(If no data, “0” is displayed. Press the
Average diastolic, or START button to turn the
Average pulse
device off.)
Average ofbutton (or the
2. Each time theall the data
button to display the data in thereverse order) is pressed, the memory
Systolic
data is displayed as follows.
Diastolic

Most recent data (No.n, in theexample, No.35)
Three seconds after the data number
display, the measurement data isdisplayed.

Pulse
Most recent data

Last data (No.1)
Three seconds after the data number
display, the measurement data isdisplayed.
3. After the last data is displayed, press
thebutton to return the average
display of all measurements.

Systolic
Diastolic
Pulse
Last data (Oldest)

English 18

4. Press the SET button to change theuser and display the memory display
for that user.
5. Press the START button to turn thedevice off.
After one minute of non-operation, thedevice will turn off automatically.

User 1

User 2User 3User 4

English 19

What Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements,
the IHB/AFib indicator will appear on the display with the measurement
values.
Note: We recommend contacting your physician if you see thisIHB/AFib indicator frequently.

What Is The AFib?
The heart contracts due to electrical signals occurring in heart and sends
blood through the body. Arterial fibrillation (AFib) occurs when the electricalsignal in the atrium becomes confused and leads to disturbances in thepulse interval. AFib can cause blood to stagnate in the heart, which caneasily create clots of blood, a cause of stroke and heart attack.

％IHB/AFib%IHB/AFib is displayed as frequency of IHB/AFib detected.
IHB/AFib can detect not only noises such as physical movement but also
an irregular heartbeat. Therefore, we recommend contacting yourphysician if %IHB/AFib level is high.
Number of detected
IHB/AFibs in memory
x 100 [%]
%IHB/AFib =Total number
Display of %IHB/AFib: %IHB/AFib is displayed when displaying average values.
(Refer to “2. Recalling the Data” in “Operation Mode”)%IHB/AFib is not displayed when the memory number is six or less.
Average value display

Level 0
%IHB/AFib=0

Level 1
%IHB/AFib=1 - 9

Level 2
%IHB/AFib=10 - 24

Not displayed

English 20

Level 3
%IHB/AFib=25 - 100

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing

Releasing air

Inflation in progress

Measurement in progress
Inflation complete

WHO Classification Indicator
Each segment of the bar indicator
corresponds to the WHO blood pressureclassification described on the next page.
WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimal: The indicator displays a segment, basedon the current data, corresponding to the
WHO classification.

English 21

Example:Moderate hypertension

Mild hypertension

High normal

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of the arteries.
Systolic pressure occurs when the heart contracts. Diastolic pressure
occurs when the heart expands. Blood pressure is measured in millimetersof mercury (mmHg). One's natural blood pressure is represented by thefundamental pressure, which is measured first thing in the morning while
one is still at rest and before eating.

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left unattended,
can cause many health problems including stroke and heart attack.
Hypertension can be controlled by altering lifestyle, avoiding stress, andwith medication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?
Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

WHO Blood Pressure Classification
Standards to assess highblood pressure, without
regard to age, have been
established by the World
Health Organization (WHO),as shown in the chart.

Blood Pressure Variations
An individual’s blood pressure
varies greatly on a daily andseasonal basis. It may varyby 30 to 50 mmHg due to various conditions during the day. In hypertensive
individuals, variations are even more pronounced. Normally, the bloodpressure rises while at work or play and falls to its lowest levels during sleep.
So, do not be overly concerned by the results of one measurement.
Take measurements at the same time every day using the procedure
English 22

described in this manual to get to know
your normal blood pressure. Regularreadings give a more comprehensive
blood pressure history. Be sure to notethe date and time when recording yourblood pressure. Consult your doctor tointerpret your blood pressure data.

Troubleshooting
ProblemPossible Reason

Nothingappears on thedisplay, evenwhen the poweris turned on.

Batteries are drained.

The cuff does
not inflate.

Battery terminals are
not in the correct
position.
Battery voltage is too low.
(LOW BATTERY mark)blinks. If the batteries are
drained completely, themark does not appear.
The cuff is not applied
properly.
You moved your arm or
body during measurement.

The device does
not measure.
The cuff position is not
Readings arecorrect.
too high or too
low.

Recommended Action
Replace all batteries with newones.
Reinstall the batteries withnegative and positive terminals
matching those indicated on thebattery compartment.
Replace all batteries with new
ones.
Apply the cuff correctly.
Make sure you remain very stilland quiet during measurement.
Sit comfortably and still.
Place your arm on a table with yourpalm facing upward and the cuff at
the same level as your heart.
If you have a very weak or
irregular heartbeat, the device
may have difficulty in determining
your blood pressure.

The value is different
Refer to “Why Measure Bloodfrom that measured at a
Pressure at Home?”.
clinic or doctor’s office.
Other
Remove the batteries.
Place them back properly andtake another measurement.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this product, as any attempt todo so will make your warranty invalid.
English 23

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem using
the troubleshooting instructions, contact the authorized dealer in your areaor our customer service department. The A&D customer service will
provide technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life.
However it is generally recommended to have the device inspected every
2 years, to ensure proper functioning and accuracy. Please contact theauthorized dealer in your area or A&D for maintenance.

Technical DataType
Measurement method
Measurement range

UA-767JP
Oscillometric measurement
Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / minute
Measurement accuracy
Pressure: ±3 mmHg
Pulse:±5%
Power supply
4 x 1.5V batteries (R6P, LR6 or AA) or
AC adapter (TB-233C) (Not included)
Number of measurements
Approx. 700 times LR6 (alkaline batteries)
Approx. 200 times R6P (manganese batteries)
With pressure value of 180 mmHg at room
temperature of 23 °C.
Classification
Internally powered ME equipment (Supplied bybatteries) / Class II (Supplied by adapter)
Continuous operation mode
Clinical test
According to ISO81060-2：2013
EMD
IEC 60601-1-2: 2014
MemoryLast 60 measurements for each of the four users
Operating conditions+10 to +40 °C / 15 to 85 %RH / 800 to 1060 hPa
Transport / Storage conditions -20 to +60 °C / 10 to 95 %RH / 700 to 1060 hPa
Dimensions
Approx. 140 [W] x 60 [H] x 105 [D] mm
WeightApprox. 255 g, excluding the batteries
Ingress protection
Device:
IP20
Applied part
Cuff
Type BF
English 24

Useful life

Device: 5 years (when used six times a day)Cuff:
2 years (when used six times a day)AC adapter: 5 years (when used six times a day)Accessory AC adapter
The adapter is to connect the device to a power source at home. Pleasecontact your local A&D dealer for purchasing. The AC adapter is required to
be inspected or replaced periodically.
TB-233C
Please contact your local A&D dealer forpurchasing.
The AC adapter is required to be inspected or
replaced periodically.
Symbols that are printed on the AC adapter
Symbols
Function / Meaning
For indoor use only
Class II device
Thermal fuse
Fuse
EC directive device label
EAC certification device label
Polarity of AC adapter plug
Accessories sold separately
Catalog Number
Cuff Size
Arm Size
Cuff
CUF-F-LA
Large adult cuff
31 cm to 45 cm
CUF-I
Wide range cuff
22 cm to 42 cm
CUF-F-A
Adult cuff
22 cm to 32 cm
Arm size: The circumference at the biceps.
AC adapter

Catalog Number
TB-233C

Plug (Outlet type)Type C

Note: Specifications are subject to change without prior notice.
IP classification is degrees of protection provided by enclosures inaccordance with IEC 60529. This device is protected against solid
foreign objects of 12 mm diameter and greater such as a finger.This
device is not protected against water.

English 25

MEMO

English 26

A&D Company, Ltd.
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119

Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
Tel: [31] (70) 345-8570
Fax: [31] (70) 346-7299

A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY
United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司
中国 上海市自由贸易试验区浦东南路855号世界广场32楼C,D室 邮编200120(32CD, World Plaza, No.855 South Pudong Road,China (Shanghai) Pilot Free Trade Zone,
200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599