Wrist Digital Blood Pressure Monitor

Model UB-525
Instruction Manual Original

EnglishInstrukcja obsługi Tłumaczenie

Polski

Návod k obsluze PřekladČeštinaNávod na obsluhu Preklad

Slovenčina

Priročnik za uporabo Prevod

Slovenščina

Használati útmutató Fordítás

MagyarΕγχειρίδιο οδηγιών ΜετάφρασηΕλληνικά

Priručnik s uputama Prijevod

HrvatskiManual de instrucțiuni TraducereRomână

Kasutusjuhend Tõlge

1WMPD4003914

Eesti

Lietošanas rokasgrāmata Tulkojums

Latviski

Naudojimo instrukcija Vertimas

Lietuvių k.

Contents

Dear Customers................................................................................................................................................... 2
Preliminary Remarks ........................................................................................................................................... 2
Precautions .......................................................................................................................................................... 2
Parts Identification ............................................................................................................................................... 4
Symbols ............................................................................................................................................................... 5
Using the Monitor................................................................................................................................................. 7
1. Installing / Changing the Batteries.............................................................................................................. 7
2. Applying the Cuff ........................................................................................................................................ 8
3. How to Take Accurate Measurements ....................................................................................................... 8
4. Measurement ............................................................................................................................................. 9
5. After Measurement ..................................................................................................................................... 9
Measurements ................................................................................................................................................... 10
Recalling the Memory Data ............................................................................................................................... 11
Deleting Data Stored in Memory ........................................................................................................................ 11
What Is The IHB/AFib Indicator? ....................................................................................................................... 12
What Is The AFib? ............................................................................................................................................. 12
Pressure Bar Indicator ....................................................................................................................................... 12
WHO Classification Indicator ............................................................................................................................. 13
About Blood Pressure ........................................................................................................................................ 13
What is Blood Pressure? .............................................................................................................................. 13
What is Hypertension and How is it Controlled? .......................................................................................... 13
Why Measure Blood Pressure at Home? ..................................................................................................... 14
WHO Blood Pressure Classification ............................................................................................................. 14
Blood Pressure Variations ............................................................................................................................ 14
Troubleshooting ................................................................................................................................................. 15
Maintenance ...................................................................................................................................................... 16
Technical Data ................................................................................................................................................... 16

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure monitor. Designed for ease of use and accuracy,
this device will facilitate your daily blood pressure regimen.
We recommend that you read through this manual carefully before using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical Products. This is made evident by themark of conformity. (0123: The reference number to the involved notified body)
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in the home healthcare environment.
This device is designed to measure blood pressure and pulse rate of people for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremes in temperature, humidity, direct
sunlight, shock or dust should be avoided.
Clean the device with a dry, soft cloth or a cloth dampened with water and a neutral detergent. Never use alcohol,
benzene, thinner or other harsh chemicals to clean the device.
Avoid tightly folding the cuff for long periods, as such treatment may shorten the life of the components.
The device is not water resistant. Prevent rain, sweat and water from soiling the device.
Measurements may be distorted if the device is used close to televisions, microwave ovens, cellular telephones,
X-ray or other devices with strong electrical fields.
Used equipment, parts and batteries are not treated as ordinary household waste, and must be disposed ofaccording to the applicable local regulations.
When reusing the device, confirm that the device is clean.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the wrist must be squeezed by the cuff hard enough to temporarily stop blood flow
through the artery. This may cause pain, numbness or a temporary red mark to the wrist. This condition will appear
especially when measurement is repeated successively. Any pain, numbness, or red marks will disappear with time.
English 2

Wireless communication device, such as home networking devices, mobile phones, cordless phones and their basestations, walkie-talkies can affect this blood pressure monitor. Therefore, a minimum distance of 30 cm should be
kept from such devices.
Measuring blood pressure too frequently may cause harm due to blood flow interference. Check that the operationof the device does not result in prolonged impairment of blood circulation, when using the device repeatedly.
Clinical testing has not been conducted on newborn infants and pregnant woman. Do not use on newborn infants or
pregnant woman.
If you have had a mastectomy, please consult a doctor before using the device.
Do not let children use the device by themselves and do not use the device in a place within the reach of infants. It
may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by mistake by infants.
Do not touch the batteries and the patient at the same time. That may result in electrical shock.
In the case of single components failure enclosure of near cuff may become hot and potentially cause malfunction.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially cause burns.
Allow the device to adapt to the surrounding environment before use (about one hour).
Do not inflate without wrapping the cuff around your wrist.

Contraindications

The following are precautions for proper use of the device.
Do not apply the device to a wrist with other medical electrical equipment attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult a doctor before using the device to avoid
medical problems.
Do not self-diagnose the measurement results and start treatment by yourself. Always consult your doctor for evaluationof the results and treatment.
Do not apply the device on a wrist with an unhealed wound.
Do not apply the device on an arm receiving an intravenous drip or blood transfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as a high-pressure oxygen chamber or an oxygen tent.
It may cause a fire or explosion.

English 3

Parts Identification
Display

button ( average, memory and change of preset parameters)

START

buttonBattery cover

Cuff

MEMORY
Average

Alkaline batteries (LR03 or AAA)

Display

Systolic pressure
Diastolic pressure

WHO classification indicatorandPressure bar indicator

IHB/AFib symbol

Heart markPulse rate

Battery indicator
English 4

Symbols

Symbols that are printed on the device caseSymbols

Function / MeaningStandby and Turn the device on
SYS

Systolic blood pressure in mmHg

DIA

Diastolic blood pressure in mmHg

PUL

Pulse per minuteBattery installation guide
Direct current

SN

Serial number
Date of manufacture
Type BF: Device and cuff are designed to provide special protection against electrical shocks.
EC directive medical device label

IP

International protection symbol
WEEE label
Manufacturer
EU-representative
Refer to instruction manual / booklet
Keep dry

English 5

Symbols that appear on the display
Symbols

Function / Meaning / Recommended Action
Appears while measurement is in progress. It blinks when the pulse is detected.
Remain as still as possible.
IHB/AFib symbol appears when an irregular heartbeat is detected.
It may light when a very slight vibration like shivering or shaking is detected.

Symbols that appear on the display (continued)
SymbolsorFunction / Meaning
Previous measurements stored in MEMORY
Average data
FULL BATTERY
The battery power indicator during measurement
LOW BATTERY
The battery is low when it blinks

Unstable blood pressure due to movement during measurement

Recommended Action

Replace all batteries with new ones
when the indicator blinks.
Take another measurement.
Remain still during measurement.

The systolic and diastolic values are within 10 mmHg of each other.
The pressure value did not increase during inflation.
Apply the cuff correctly, and take
The cuff is not applied correctly.
another measurement.
PUL.DISPLAY ERROR
The pulse is not detected correctly.
Remove the batteries and press the
START button, and then install the
Blood pressure monitor internal error
batteries again. If the error still
appears, contact the dealer.
English 6

Using the Monitor

1. Installing / Changing the Batteries
1. Remove the battery cover.

2. Remove the used batteries and insert new batteries into thebattery compartment as shown, taking care that the polarities(+ and -) are correct. Use only LR03 or AAA batteries.

Battery cover

3. Attach the battery cover.
Used batteriesInsert the batteries as shown in the battery compartment.
New batteries
If installed incorrectly, the device will not work.
When(LOW BATTERY mark) blinks on the display,replace all batteries with new ones. Do not mix old and newbatteries. It may shorten the battery life, or cause the device
to malfunction.
(LOW BATTERY mark) does not appear when thebatteries are drained.
The battery life varies with the ambient temperature and may be shorter at low temperatures. Generally, two new
LR03/AAA batteries will last approximately for four months when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device are for testing monitor performance andmay have a limited life.
Remove the batteries if the device is not to be used for a long period of time. The batteries may leak and cause a
malfunction.

English 7

2. Applying the Cuff
1. Wrap the cuff around your wrist about 1 cm above your hand as shown in the figure
at the right.

About 1 cm

2. Apply the cuff tightly using the Velcro strip.
Note: For accurate measurements, apply the cuff tightly and measure on a bare wrist.

3. How to Take Accurate Measurements

For the most accurate blood pressure measurement:
Remain still and keep quiet during measurement.
Sit down in a comfortable position. Place your elbow on a table with your palm facing upward and the cuff is at thesame level as your heart.
Relax for about five to ten minutes before measurement. If you are excited or depressed by emotional stress, themeasurement will reflect this stress as a higher (or lower) than normal blood pressure reading and the pulse
reading will usually be faster than normal.
Try to measure your blood pressure at about the same time every day.
An individual's blood pressure varies constantly, depending on what they are doing, what they have eaten and what
they drink can have a very strong and rapid effect on your blood pressure.
Do not measure immediately after physical exercise or a bath. Rest for twenty or thirty minutes before taking themeasurement.
Do not cross your legs. Keep your feet flat on the floor and straighten your back.
This device bases its measurements on the heartbeat. If you have a very weak or irregular heartbeat, the device
may have difficulty determining your blood pressure.
Should the device detect a condition that is abnormal, it will stop the measurement and display an error symbol.
See page 6 for the description of symbols.
This blood pressure monitor is intended for use by adults. Consult with your physician before using this device on achild. A child should not use this device unattended.
The automatic blood pressure monitor's performance may be affected by excessive temperature or humidity, or
altitude.
English 8

4. Measurement

During measurement, it is normal for the cuff to feel very tight.

5. After Measurement

While readings are displayed, if you press the START button to turn the device off, new readings are stored inmemory.
While readings are displayed, if you press the
button to turn the device off, new readings are not stored.
Remove the cuff and record your data.
Notes: The device is provided with an automatic power shut-off function which stores the current data in thememory and turns the device off automatically one minute after measurement. Allow at least three minutes
between measurements on the same person.

English 9

Measurements
Note: The UB-525, once used, will provide an inflation appropriate to the user.
1. Wrap the cuff around your wrist. Sit comfortably with the cuff at the same levelas your heart and relax.
2. Press the START button. All of the display segments are displayed.
3. Zero (0) is displayed blinking briefly. Then the display changes, as measurement
begins. The cuff starts to inflate. It is normal for the cuff to feel very tight. The
measurement starts automatically when inflation starts, and the(heart mark)blinks.
Note: If you wish to stop inflation at any time, press the START button again.
4. When the measurement is complete, the device displays readings (of thesystolic and diastolic pressure readings, pulse rate, WHO classification, I.H.B.
symbol). The cuff exhausts the remaining air and completely deflates
automatically.
Note: If you do not want to store new readings in memory, press thewhile readings are displayed.

button5. Press the START button again to turn the device off. Remove the cuff.
Notes: The device is provided with an automatic power shut-off function.
Allow at least three minutes between measurements on the same person.

English 10

All segments
Zero display
Inflation is started
Inflating and
Measurement inprogress
Systolic pressure
Diastolic pressure
WHO classification
I.H.B.symbol
Pulse rate

Recalling the Memory Data
Note: This device stores the last 60 measurements in memory.
Memory mode
button when turning the device off.
1. Press the
The average of all measurements and the number of data are displayed.
If no data, “0” is displayed. Press theor START button to turn the
Average of all the data
device off.
Number of all data
2. Use the following buttons to display data (of number and measurement data).
The device displays the average of all measurements and thenumber of data are displayed.
Ex.: No.30
Pressing the
button each time , the device displays as follows:
Most recent data
Average data of all measurements taken between.
Data (of number and measurement data).
The device displays in order from most recent data.
Last data (Oldest)
In the example: No.30 & data
No.29 & data
No.01 & data.
button after oldest data is displayed, the device
3. If you press theproceeds to step 1, the average of all measurements and the number ofdata are displayed.
4. Press the START button to turn the device off.
After one minute of non-operation, the device will turn off automatically.

Deleting Data Stored in Memory
1. Press and hold the

button until turning off the device automatically.

The device displays themark, deletes data stored in memory while blinking
themark and turns off automatically.
English 11

Turn off
buttonbuttonbuttonbuttonTurn off
buttonWhat Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements, the IHB/AFib indicator will appear on thedisplay with the measurement values.
Note: We recommend contacting your physician if you see thisIHB/AFib indicator frequently.

What Is The AFib?

The heart contracts due to electrical signals occurring in heart and sends blood through the body. Atrial fibrillation(AFib) occurs when the electrical signal in the atrium becomes confused and leads to disturbances in the pulseinterval. AFib can cause blood to stagnate in the heart, which can easily create clots of blood, a cause of stroke andheart attack.
Pressurizing

Pressure Bar Indicator

The indicator monitors the progress of pressure during
measurement.
Inflation in progress
Measurement in progress
Inflation complete

English 12

WHO Classification Indicator
Each six segments of the bar indicator correspond to the
WHO blood pressure classification is described on page 14.
ExampleModerate
hypertension

Mild hypertension

High normal

WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimal: The indicator displays a segment,
based on the current data,
corresponding to the WHO classification.

What is Blood Pressure?

About Blood Pressure

Blood pressure is the force exerted by blood against the walls of the arteries. Systolic pressure occurs when theheart contracts. Diastolic pressure occurs when the heart expands. Blood pressure is measured in millimeters of
mercury (mmHg). One's natural blood pressure is represented by the fundamental pressure, which is measured
first thing in the morning while one is still at rest and before eating.

What is Hypertension and How is it Controlled?

Hypertension, an abnormally high arterial blood pressure, if left unattended can cause many health problems
including stroke and heart attack. Hypertension can be controlled by altering lifestyle, avoiding stress, and withmedication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight
English 13

Why Measure Blood Pressure at Home?

Blood pressure measured at a clinic or doctor's office may cause apprehension and can produce an elevatedreading, 25 to 30 mmHg higher than that measured at home. Home measurement reduces the effects of outside
influences on blood pressure readings, supplements the doctor's readings and provides a more accurate, complete
blood pressure history.

WHO Blood Pressure Classification

Diastolic blood pressureStandards to assess high blood pressure, without regard to age, have Reference Material: Journal of Hypertension 1999, Vol 17 No.2
mmHg
Grade 3 hypertension (severe)
been established by the World Health Organization (WHO), as shown110
in the chart at the right.
105
Grade 2 hypertension (moderate)

Blood Pressure Variations

100
95
90
85
80

Grade 1 hypertension (mild)

Blood pressure

An individual’s blood pressure varies greatly on a daily and seasonalHigh-normal
basis. It may vary by 30 to 50 mmHg due to various conditions during
Normalthe day. In hypertensive individuals variations are even more
Optimalpronounced. Normally, the blood pressure rises while at work or play120 130 140 150 160 170 180
Systolic blood pressure
mmHg
and falls to its lowest levels during sleep. So, do not be overly
concerned by the results of one measurement.
Typical fluctuation within a day
Take measurements at the same time every day using the procedure described in this(Measured every five minutes)
manual to get to know your normal blood pressure. Regular readings give a more
Systolic
comprehensive blood pressure history. Be sure to note the date and time whenrecording your blood pressure. Consult your doctor to interpret your blood pressure
data.

PM

English 14

Sleep

DiastolicAM

TimeTroubleshooting
ProblemPossible Reason
Nothing appears in the Batteries are drained.
display, even when
Battery terminals are not in the correct
the device is turned
position.
on.
Battery voltage is too low.
(LOW BATTERY mark) blinks.
The cuff does not
inflate.
If the batteries are drained completely,
the mark does not appear.
The cuff is not applied properly.
You moved your wrist or body during themeasurement.
The device does notmeasure.
Readings are too high The cuff position is not correct.
or too low.

Other

Recommended Action
Replace all batteries with new ones.
Reinstall the batteries with negative and positive
terminals matching those indicated in the battery
compartment.
Replace all batteries with new ones.

Apply the cuff correctly.
Make sure you remain very still and quiet during
the measurement.
Sit comfortably and still.
Place your arm on a table with your palm facing
upward and the cuff at the same level as yourheart.
If you have a very weak or irregular heartbeat,the device may have difficulty in determining yourblood pressure.
The value is different from that measured See the section “Why Measure Blood Pressure
at a clinic or doctor’s office.
at Home?”.
Remove the batteries. Place them back properly
and try measurement again.

Note: If the actions described above do not solve the problem, contact the dealer. Do not attempt to open or repair
this product yourself, as any attempt to do so will make your warranty invalid.
English 15

MaintenanceDo not open the device. It uses delicate electrical components and an intricate air unit that could be damaged. If you
cannot fix the problem using the troubleshooting instructions, contact the authorized dealer in your area or ourcustomer service department. The A&D customer service will provide technical information, spare parts and units toauthorized dealers.
The device was designed and manufactured for a long service life. However it is generally recommended to havethe device inspected every 2 years, to ensure proper functioning and accuracy. Please contact the authorized
dealer in your area or A&D for maintenance.

Technical DataType
Measurement method
Measurement range

Measurement accuracy
Power supply
Number of measurements
Wrist circumference
Classification
Applied part
Useful life

UB-525
Oscillometric measurement
Pressure:0 – 299 mmHg
Systolic pressure: 60 – 279 mmHg
Diastolic pressure: 40 – 200 mmHg
Pulse:
40 – 180 beats / minutePressure: ±3 mmHg
Pulse:±5 %2 x 1.5 V alkaline batteries (LR03 or AAA)Approx. 250 measurements, when AAA alkaline batteries are used, with pressure valueof 170 mmHg at room temperature of 23 °C.
13.5 – 21.5 cm
Internally powered ME equipment (Continuous operation mode)
Cuff Type BF
Device: 5 years (when used six times a day)English 16

Clinical test
EMD
MemoryOperating conditions
Transport / storage conditions
Dimensions
WeightIngress protection

According to ISO81060-2：2013
In the clinical validation study, K5 was used on 85 subjects for determination ofdiastolic blood pressure.
IEC 60601-1-2: 2014
Last 60 measurements+10 to +40 °C / 15 to 85 %RH / 800 to 1060 hPa-20 to +60 °C / 10 to 95 %RH / 700 to 1060 hPa
Approx. 56 [W] x 88 [H] x 21.5 [D] mm
Approx. 90 g, excluding the batteries
IP20

Note: Specifications are subject to change for improvement without prior notice.
IP classification is the degrees of protection provided by enclosures in accordance with IEC 60529. This device isprotected against solid foreign objects of 12 mm diameter and greater such as a finger. This device is not protectedagainst water.

English 17

MEMO

English 18

Spis treści

Szanowny kliencie ............................................................................................................................................... 2
Uwagi wstępne .................................................................................................................................................... 2Środki ostrożności ............................................................................................................................................... 2
Oznaczenie części ............................................................................................................................................... 4
Symbole ............................................................................................................................................................... 5
Korzystanie z ciśnieniomierza ............................................................................................................................. 7
1. Wkładanie/wyjmowanie baterii ................................................................................................................... 7
2. Zakładanie mankietu .................................................................................................................................. 8
3. Wykonywanie dokładnych pomiarów ......................................................................................................... 8
4. Pomiar ........................................................................................................................................................ 9
5. Po pomiarze ............................................................................................................................................... 9
Pomiary.............................................................................................................................................................. 10
Wywoływanie zapisanych danych ..................................................................................................................... 11
Usuwanie wszystkich danych z pamięci ............................................................................................................ 11
Czym jest wskaźnik IHB/AFib? .......................................................................................................................... 12
Czym jest migotanie przedsionków (AFib)? ....................................................................................................... 12Wskaźnik słupkowy ciśnienia ............................................................................................................................. 12Wskaźnik klasyfikacji WHO ............................................................................................................................... 13
Informacje o ciśnieniu tętniczym ........................................................................................................................ 13
Czym jest ciśnienie tętnicze? ....................................................................................................................... 13
Czym jest nadciśnienie i jak można je kontrolować? ................................................................................... 13
Dlaczego warto mierzyć ciśnienie tętnicze w warunkach domowych? ......................................................... 14
Klasyfikacja ciśnienia tętniczego wg WHO................................................................................................... 14
Zmiany ciśnienia tętniczego ......................................................................................................................... 14
Rozwiązywanie problemów ............................................................................................................................... 15
Konserwacja ...................................................................................................................................................... 16
Dane techniczne ................................................................................................................................................ 16

Polski1

Szanowny kliencie
gratulujemy zakupu ciśnieniomierza A&D. Łatwość obsługi i wysoka dokładność urządzenia pozwolą na skuteczne monitorowanieciśnienia tętniczego.
Przed pierwszym użyciem urządzenia zalecamy dokładne przeczytanie niniejszej instrukcji.

Uwagi wstępne
Niniejsze urządzenie spełnia wymagania Dyrektywy Rady 93/42 EWG dotyczącej wyrobów medycznych. Jest to potwierdzone
znakiem zgodności
. (0123: numer referencyjny jednostki notyfikowanej).
Urządzenie jest przeznaczone do użycia przez osoby dorosłe — zabrania się dokonywania pomiarów ciśnienia u noworodków
i niemowląt.
Środowisko użytkowania. Urządzenie jest przeznaczone do obsługi samodzielnej w warunkach domowych.
Niniejsze urządzenie jest przeznaczone do pomiarów ciśnienia tętniczego i tętna dla celów diagnostycznych.

Środki ostrożności
Do konstrukcji tego urządzenia użyto precyzyjnych podzespołów. Należy chronić je przed działaniem skrajnych temperatur, wilgoci,
bezpośredniego światła słonecznego, uderzeń oraz kurzu.
Urządzenie należy czyścić miękką, suchą ściereczką lub ściereczką nasączoną wodą i neutralnym detergentem. Do czyszczeniaurządzenia nigdy nie należy używać alkoholu, benzenu, rozcieńczalników ani innych środków chemicznych o agresywnym działaniu.
Nie należy składać mankietu zbyt ciasno na dłuższy czas, gdyż może to spowodować zmniejszenie jego trwałości.
Urządzenie nie jest wodoodporne. Unikać kontaktu urządzenia z deszczem, potem i wodą.
W przypadku korzystania z urządzenia w pobliżu telewizorów, kuchenek mikrofalowych, telefonów komórkowych, aparatów
rentgenowskich i innych urządzeń o silnym polu elektrycznym wyniki pomiaru mogą być zaburzone.
Zużyte urządzenie, części oraz baterie nie mogą być traktowane jak normalne odpady z gospodarstwa domowego i należy jezutylizować zgodnie z obowiązującymi lokalnymi przepisami.
Przed ponownym użyciem urządzenia należy upewnić się, że jest ono czyste.
Urządzenia nie należy poddawać modyfikacjom. Może to stać się przyczyną wypadków lub doprowadzić do uszkodzenia urządzenia.
Aby dokonać pomiaru ciśnienia tętniczego, należy na tyle mocno ścisnąć nadgarstek mankietem, by chwilowo zatrzymać przepływ
krwi przez tętnicę. Może to spowodować ból, drętwienie lub tymczasowe zaczerwienienie nadgarstka. Taki stan może pojawiać się
zwłaszcza w sytuacjach, gdy pomiar jest kolejno powtarzany. W miarę upływu czasu objawy zanikają.

Polski2

Na działanie monitora ciśnienia tętniczego mogą mieć także wpływ bezprzewodowe urządzenia komunikacyjne, takie jak urządzenia
działające w sieci domowej, telefony komórkowe, telefony bezprzewodowe i ich stacje bazowe oraz krótkofalówki. Zalecane jest
utrzymanie odległości ok. 30 cm od takich urządzeń.
Zbyt częste pomiary ciśnienia tętniczego mogą być szkodliwe z uwagi na zakłócenie przepływu krwi. Należy upewnić się, że wielokrotneużywanie urządzenia nie będzie skutkowało przedłużonym zaburzeniem przepływu krwi.
Nie przeprowadzono badań klinicznych u noworodków ani kobiet w ciąży. Urządzenia nie należy używać do pomiaru ciśnienia krwi
u noworodków i kobiet w ciąży.
Po zabiegu mastektomii przed użyciem urządzenia należy skontaktować się z lekarzem.
Nie wolno pozwalać dzieciom na samodzielne korzystanie z urządzenia ani nie należy używać urządzenia w pobliżu niemowląt. Możeto stać się przyczyną wypadków lub doprowadzić do uszkodzenia mienia.
Urządzenie zawiera małe części, które w przypadku połknięcia przez niemowlęta stwarzają ryzyko udławienia się.
Nie należy dotykać jednocześnie baterii i ciała pacjenta. Może to spowodować porażenie prądem elektrycznym.
Jeśli dojdzie do uszkodzenia pojedynczego podzespołu obudowa przy mankiecie może się nagrzać i spowodować nieprawidłowe
działanie.
Używanie akcesoriów niewymienionych w niniejszej instrukcji może obniżyć poziom bezpieczeństwa.
W przypadku zwarcia baterii może ona nagrzać się do wysokiej temperatury, stwarzając zagrożenie poparzeniem.
Przed użyciem należy pozostawić urządzenie na około godzinę, aby umożliwić jego dostosowanie się do warunków otoczenia.
Nie nadmuchiwać mankietu niezałożonego na nadgarstek.

PrzeciwwskazaniaPoniżej opisano środki ostrożności dotyczące właściwego korzystania z urządzenia.
Nie zakładać mankietu na nadgarstek, na który założono inne urządzenia medyczne. Urządzenia te mogą nie funkcjonować prawidłowo.
Osoby, które mają poważne problemy krążeniowe w ramieniu powinny przed użyciem urządzenia skonsultować się z lekarzem, abyzapobiec problemom natury medycznej.
Nie należy samodzielnie dokonywać diagnozy ani podejmować leczenia w oparciu o uzyskane wyniki. W celu oceny wyników i wyboru
sposobu leczenia należy zawsze konsultować się z lekarzem.
Nie zakładać urządzenia na nadgarstek, na którym znajduje się niezagojona rana.
Nie zakładać urządzenia na ramię, do którego podłączony jest wlew dożylny lub do którego dokonywana jest transfuzja krwi. Może tostać się przyczyną odniesienia obrażeń lub wypadków.
Nie używać urządzenia w obecności łatwopalnych gazów jak np. gaz znieczulający. Może być to przyczyną wybuchu.
Nie używać urządzenia w warunkach wysokiego stężenia tlenu, takich jak wysokociśnieniowe komory tlenowe lub namioty tlenowe.
Może to stać się przyczyną pożaru lub wybuchu.

Polski3

Oznaczenie części
Wyświetlacz

Przycisk(wartość średnia, pamięć i zmiana zadanych parametrów)

Przycisk START
Pokrywa komory baterii

MankietPAMIĘĆ
Wartość uśredniona

Baterie alkaliczne (LR03 lub AAA)Wyświetlacz
Ciśnienie skurczoweCiśnienie rozkurczoweWskaźnik klasyfikacji WHO
orazwskaźnik słupkowy ciśnienia

Symbol IHB/AFibWskaźnik sercaTętnoWskaźnik poziomu
naładowania baterii
Polski4

Symbole na obudowie urządzenia

SymboleSymboleFunkcja/znaczenieWłączanie i wyłączanie urządzenia
SYS

Skurczowe ciśnienie tętnicze w mmHg

DIA

Rozkurczowe ciśnienie tętnicze w mmHg

PUL

Uderzenia na minutę
Wskazówka dotycząca montażu bateriiPrąd stały

SN

Numer seryjny
Data produkcji
Typ BF: urządzenie i mankiet zaprojektowano tak, aby zapewnić specjalną ochronę przed porażeniemprądem elektrycznym.
Etykieta dyrektywy WE odnośnie urządzeń medycznychIPMiędzynarodowy symbol ochrony urządzenia
Etykieta WEEE
ProducentPrzedstawiciel w UE
Przeczytać instrukcję obsługi/broszurę informacyjną
Przechowywać w suchym miejscu
Polski5

Symbole widoczne na wyświetlaczu
SymboleFunkcja/znaczenie/zalecane działanie
Symbol widoczny podczas pomiaru. Zaczyna migać po wykryciu tętna.
Unikać ruchów.
Symbol IHB/AFib (nieregularna praca serca/migotanie przedsionków) Pojawia się, gdy zostaje wykryte
nieregularne bicie serca.
Może wyświetlić się, gdy zostanie wykryta bardzo łagodna wibracja jak np. drżenie lub dreszcze.

Symbole widoczne na wyświetlaczu (ciąg dalszy)
Symbolelub

Funkcja/znaczenie
Poprzednie pomiary zapisane w PAMIĘCI
Wartość uśredniona danych
BATERIA NAŁADOWANA
Wskaźnik poziomu naładowania baterii podczas pomiaru
NISKI POZIOM BATERIIMiganie oznacza niski stopień naładowania baterii
Niestabilne ciśnienie tętnicze spowodowane poruszaniem się
podczas pomiaruRóżnica pomiędzy wartością skurczową a rozkurczową zawiera
się w przedziale 10 mmHg.
Wartość ciśnienia nie zwiększyła się podczas nadmuchiwania.
Mankiet nie został zapięty prawidłowo.
BŁĄD TĘTNA
Nie wykryto prawidłowego tętna.

Zalecane działanie

Gdy ten wskaźnik będzie migać,należy wymienić baterie.
Należy wykonać kolejny pomiar.
Pozostawać w bezruchu.

Założyć mankiet w prawidłowy sposób
i ponowić pomiar.

Wyjąć baterie i nacisnąć przycisk
START , a następnie ponownie włożyć
baterie. Jeżeli błąd będzie
występował nadal, należy
skontaktować się ze sprzedawcą.

Wewnętrzny błąd ciśnieniomierza

Polski6

Korzystanie z ciśnieniomierza

1. Wkładanie/wyjmowanie baterii
1. Zdjąć pokrywę komory baterii.

2. Wyjąć zużyte baterie i włożyć w ich miejsce nowe bateriew przedstawiony sposób, zwracając uwagę na polaryzację (+ i -).
Stosować tylko baterie LR03 lub AAA.

Pokrywa komorybaterii

3. Założyć pokrywę komory baterii.

OSTROŻNIE

Zużyte baterie

Baterie należy umieszczać w komorze w sposób
przedstawiony na ilustracji.
Nowe baterie
W przypadku nieprawidłowego włożenia baterii urządzenie
nie włączy się.
Gdy na wyświetlaczu zacznie migać symbol(NISKIPOZIOM BATERII), należy wymienić wszystkie baterie na
nowe. Nie używać jednocześnie baterii zużytych i nowych.
Może to spowodować skrócenie ich żywotności lub
nieprawidłowe działanie urządzenia.
Symbol(NISKI POZIOM BATERII) nie będzie wyświetlany,jeśli baterie ulegną całkowitemu rozładowaniu.
Żywotność baterii zależy od temperatury otoczenia i może być mniejsza w niższych temperaturach. Zasadniczo
dwie nowe baterie LR03/AAA powinny wystarczyć na ok. cztery miesiące w przypadku dokonywania dwóch
pomiarów każdego dnia.
Należy używać wyłącznie baterii wymienionych w instrukcji. Baterie dostarczone wraz z urządzeniem służą
wyłącznie do sprawdzenia działania urządzenia i mogą mieć ograniczoną żywotność.
Jeżeli urządzenie nie będzie używane przez długi czas, należy wyjąć baterie. Baterie mogą wyciec i spowodować
awarię urządzenia.
Polski 7

2. Zakładanie mankietu

1. Owinąć mankiet wokół nadgarstka około 1 cm nad dłonią tak, jak toprzedstawiono na ilustracji po prawej stronie.
2. Owinąć mankiet ciasno wokół nadgarstka i zapiąć go za pomocą rzepu.
Uwaga: aby uzyskać dokładny pomiar, należy owinąć mankiet ciasno wokółnadgarstka i dokonywać pomiaru na nieosłoniętym nadgarstku.

3. Wykonywanie dokładnych pomiarów

Ok. 1 cm

Aby uzyskać najdokładniejszy pomiar ciśnienia tętniczego, należy:Nie poruszać się ani nie rozmawiać podczas pomiaru.
Należy usiąść w wygodnej pozycji. Oprzeć łokieć na stole z dłonią skierowaną ku górze i mankietem założonym na
wysokości serca.
Przed pomiarem zaleca się odprężyć przez od pięciu do dziesięciu minut. U osób podekscytowanych lub
przygnębionych w wyniku stresu emocjonalnego pomiar odzwierciedli te stany jako wyższe (lub niższe) ciśnienietętnicze niż w stanie normalnym, a wartość tętna będzie zwykle wyższa niż normalnie.
Pomiary należy wykonywać w miarę możliwości codziennie o podobnej porze dnia.
Ciśnienie krwi podlega ciągłym zmianom w zależności od wykonywanych czynności; spożyte pokarmy i wypite
napoje mogą istotnie i gwałtownie wpływać na ciśnienie tętnicze.
Nie dokonywać pomiarów bezpośrednio po ćwiczeniach fizycznych lub po kąpieli. Przed pomiarem należy
odpocząć od dwudziestu do trzydziestu minut.
Nie krzyżować nóg. Oprzeć stopy na podłodze i wyprostować się.
Urządzenie opiera swój pomiar na tętnie. Jeżeli tętno jest bardzo słabe lub nieregularne, urządzenie może miećproblemy z określeniem ciśnienia tętniczego.
Jeżeli urządzenie wykryje stan odbiegający od normy, wówczas pomiar zostanie przerwany i wyświetli się
komunikat o błędzie. Opis symboli zamieszczono na stronie 6.
Ten monitor ciśnienia tętniczego jest przeznaczony wyłącznie dla osób dorosłych. Przed zastosowaniemurządzenia u dziecka należy skonsultować się z lekarzem. Dzieci nie mogą używać urządzenia bez nadzoru.
Na działanie automatycznego monitora ciśnienia tętniczego może wpływać nadmierna temperatura, wilgotność lub
wysokość nad poziomem morza.
Polski 8

4. Pomiar

Podczas pomiaru mankiet może uciskać ramię — jest to zjawisko normalne.

5. Po pomiarzeNaciśnięcie przycisku START w celu wyłączenia urządzenia po dokonaniu pomiaru spowoduje zapisanie
zmierzonych wartości w pamięci.
w celu wyłączenia urządzenia po dokonaniu pomiaru spowoduje natomiast, że zmierzoneNaciśnięcie przycisku
wartości nie zostaną zapisane.
Zdjąć mankiet i zapisać dane.
Uwaga: urządzenie posiada funkcję samoczynnego wyłączania, która zapisuje zmierzoną wartość w pamięci
i wyłącza urządzenie po około minucie od zakończenia pomiaru. W razie wykonywania pomiaru na tej
samej osobie należy odczekać co najmniej trzy minuty między kolejnymi pomiarami.

Polski9

Pomiary
Uwaga: po użyciu urządzenie UB-525 zostanie nadmuchane odpowiednio dla danegoużytkownika.
1. Owinąć mankiet wokół nadgarstka. Usiąść wygodnie, umieścić mankiet na
poziomie serca i zrelaksować się.
2. Nacisnąć przycisk START . Wyświetlą się wszystkie segmenty wyświetlacza.

Wszystkie segmenty

3. Wyświetli się cyfra zero (0) i przez krótki czas będzie migać. Następnie
wskazane wyświetlacza zmieni się, a pomiar zostanie rozpoczęty. Ciśnienie
w mankiecie zacznie rosnąć. Mankiet może bardzo uciskać ramię — jest to
zjawisko normalne. Pomiar zostanie uruchomiony automatycznie po
rozpoczęciu nadmuchiwania — znacznik(serca) zacznie migać.

Wskazanie zerowe
Rozpoczęcie
nadmuchiwania

Uwaga: aby zatrzymać nadmuchiwanie mankietu w dowolnej chwili, należy
ponownie nacisnąć przycisk START .

Nadmuchiwanie
i pomiar w toku

4. Po zakończeniu pomiaru wyświetlą się zmierzone wartości (ciśnienia
skurczowego i rozkurczowego, tętna, klasyfikacji WHO oraz symbol IHB).
Z mankietu zostanie automatycznie usunięte powietrze.

Ciśnienie skurczoweCiśnienie rozkurczowe
Klasyfikacja WHO
Uwaga: aby nie zapisać zmierzonych wartości w pamięci, należy podczas ich
Symbol IHBTętnowyświetlania nacisnąć przycisk .

5. Ponownie nacisnąć przycisk START , aby wyłączyć urządzenie. Zdjąć
mankiet.
Uwaga: urządzenie posiada funkcję samoczynnego wyłączania.
W razie wykonywania pomiaru na tej samej osobie należy odczekać conajmniej trzy minuty między kolejnymi pomiarami.
Polski10

Wywoływanie zapisanych danych

Uwaga:
urządzenie może zapisać dane maksymalnie 60 pomiarów.
1. Nacisnąć przycisk
podczas wyłączania urządzenia.
Wyświetlą się uśredniona wartość ze wszystkich pomiarów oraz liczba
zapisanych danych. W razie braku zapisanych danych wyświetli się cyfra
lub START , aby wyłączyć urządzenie.
„0”. Nacisnąć przycisk
2. Aby wyświetlić dane (liczbę pomiarów i zmierzone wartości), należy użyćnastępujących przycisków.
Urządzenie wyświetli uśrednioną wartość ze wszystkich pomiarów
oraz liczbę zapisanych danych.
Każdorazowe naciśnięcie przycisku
spowoduje, że urządzeniewyświetli:
Wartość uśredniona ze wszystkich pomiarów
Dane (liczbę zapisanych danych i zmierzone wartości).
Urządzenie wyświetla dane począwszy od najnowszych.
Przykład: nr 30 i danenr 29 i danenr 01 i dane.
po wyświetleniu najstarszych danych
3. Naciśnięcie przycisku
spowoduje, że urządzenie przejdzie do kroku 1. — wyświetlą sięuśrednione dane ze wszystkich pomiarów oraz liczba zapisanych danych.
4. Nacisnąć przycisk START , aby wyłączyć urządzenie.
Po minucie bezczynności urządzenie wyłączy się automatycznie.

Tryb pamięci

Przycisk
Wartość uśredniona
wszystkich danych
Liczba wszystkich
zapisanych danych

Przycisk
Najstarsze dane

1. Nacisnąć i przytrzymać przycisk , aż urządzenie wyłączy się samoczynnie.
Wyświetli się znacznik; następnie znacznik
zacznie migać — wszystkie dane
zapisane w pamięci zostaną usunięte, po czym urządzenie wyłączy się samoczynnie.
11

Przycisk

Np. 30Najnowsze dane

Usuwanie wszystkich danych z pamięci

PolskiWyłączenie

PrzyciskWyłączenie
Przycisk

Czym jest wskaźnik IHB/AFib?
Gdy urządzenie wykryje nieregularne bicie serca podczas pomiaru, na wyświetlaczu pojawi się wskaźnik IHB/AFibwraz ze zmierzonymi wartościami.
Uwaga: jeśli wskaźnik IHB/AFibbędzie pojawiał się często, zalecamy skonsultowanie się z lekarzem.

Czym jest migotanie przedsionków (AFib)?

Serce kurczy się pod wpływem bodźców elektrycznych powstających w komórkach serca, pompując krew.
Migotanie przedsionków (Arterial fibrillation, AFib) występuje, gdy bodziec elektryczny w przedsionku ulega
zaburzeniom, co z kolei prowadzi do nieregularności pracy serca. Migotanie przedsionków (AFib) może być
przyczyną wstrzymania obiegu krwi w sercu, co może doprowadzić do powstania zakrzepów, a w konsekwencji do
udaru lub zawału serca.
NadmuchiwanieWskaźnik słupkowy ciśnienia

Ten wskaźnik monitoruje zmiany ciśnienia podczas
pomiaru.
Nadmuchiwanie w toku
Pomiar w toku
Nadmuchiwanie zakończone

Polski12

Wskaźnik klasyfikacji WHOKażdy z sześciu segmentów wskaźnika słupkowego
odpowiada klasyfikacji ciśnienia tętniczego WHO opisanej na
stronie 14.
Przykład
Nadciśnienie
Nadciśnienie
Prawidłowe
umiarkowanełagodnewysokie ciśnienieWskaźnik klasyfikacji WHO
Nadciśnienie ostre
Nadciśnienie umiarkowane
Nadciśnienie łagodne
Prawidłowe wysokie ciśnienie
Normalne ciśnienie
Optymalne ciśnienie: Na podstawie bieżących danychwskaźnik wyświetla segment
odpowiadający klasyfikacji WHO.

Informacje o ciśnieniu tętniczym

Czym jest ciśnienie tętnicze?

Ciśnienie tętnicze jest siłą wywieraną przez krew na ścianki tętnic. Ciśnienie skurczowe występuje podczas kurczenia
się przedsionków serca. Ciśnienie rozkurczowe występuje natomiast podczas rozkurczania się przedsionków.
Ciśnienie tętnicze wyraża się w milimetrach słupa rtęci (mmHg). Naturalne ciśnienie tętnicze jest przedstawiane jakociśnienie bazowe, mierzone bezpośrednio po obudzeniu się, w stanie spoczynku, przed posiłkiem.

Czym jest nadciśnienie i jak można je kontrolować?

Nadciśnienie jest nienormalnie wysokim ciśnieniem tętniczym, które, w przypadku braku kontroli, może prowadzić
do wielu problemów zdrowotnych, takich jak udar lub zawał serca. Nadciśnienie można kontrolować, zmieniając
nawyki, unikając stresu oraz przyjmując leki pod kontrolą lekarza.
Stosowanie się do poniższych zaleceń pozwoli kontrolować nadciśnienie lub zapobiec jego wystąpieniu:
Nie palić tytoniu
Regularnie ćwiczyć
Zmniejszyć spożycie soli i tłuszczu
Poddawać się regularnym badaniom lekarskim
Dbać o prawidłową masę ciała
Polski 13

Dlaczego warto mierzyć ciśnienie tętnicze w warunkach domowych?

Światowa Organizacja Zdrowia (World Health Organization,WHO) ustaliła normy oceny ciśnienia tętniczego —przedstawiono je na wykresie po prawej stronie.

Zmiany ciśnienia tętniczegoRóżnice poziomu ciśnienia tętniczego mogą być znaczne —zarówno w ciągu dnia, jak i dłuższego okresu. Różnica dziennamoże wynosić od 30 do 50 mmHg. U osób z nadciśnieniem różnice
te są jeszcze wyraźniejsze. Zwykle ciśnienie wzrasta podczas pracy
lub aktywności i spada do najniższego poziomu podczas snu. Wynik
jednego pomiaru nie powinien być zatem przyczyną obaw.
Pomiarów należy dokonywać o tej samej porze dnia, postępując
zgodnie ze wskazówkami zamieszczonymi w niniejszej instrukcji.
Pozwoli to określić normalny poziom ciśnienia tętniczego. Regularne
pomiary pozwolą uzyskać dokładną historię wartości ciśnieniatętniczego. Za każdym razem należy odnotowywać godzinę i datępomiaru. Aby prawidłowo zinterpretować wskazania, należy
skonsultować się z lekarzem.

Ciśnienie tętnicze

Klasyfikacja ciśnienia tętniczego wg WHO

Rozkurczowe ciśnienie tętnicze

Pomiar ciśnienia tętniczego w przychodni lub gabinecie lekarskim może wskazywać podwyższone wartości, któremogą być o 25–30 mmHg wyższe niż wskazane podczas pomiaru w warunkach domowych. Pomiar w warunkachdomowych może ograniczyć wpływ czynników zewnętrznych na ciśnienie, a ponadto może uzupełniać pomiary
dokonywane w gabinecie lekarskim i pozwolić na uzyskanie dokładniejszej i kompletnej historii pomiarów.
Materiał referencyjny: Journal ofHypertension 1999, tom 17 nr 2
Nadciśnienie 3. poziomu (ostre)
Nadciśnienie 2. poziomu (umiarkowane)
Nadciśnienie 1. poziomu (łagodne)
Prawidłowe wysokie ciśnienie

Normalne ciśnienie
Optymalneciśnienie

Skurczowe ciśnienie tętnicze
Typowe wahania w ciągu dnia
(pomiar co pięć minut)
Skurczowe

Rozkurczowe
popołudnie

Polski14

Sen
przedpołudnie

CzasRozwiązywanie problemów
ProblemPrawdopodobna przyczynaWyświetlacz pozostaje Baterie są rozładowane.
pusty mimo włączenia Bieguny baterii nie są ustawioneurządzenia.
prawidłowo.
Zbyt niskie napięcie baterii.
(NISKI POZIOM
SymbolMankiet nie
BATERII) miga.
napompowuje się.
W przypadku całkowitego rozładowania
baterii wskaźnik ten nie pojawi się.
Mankiet nie został prawidłowo zapięty.
Podczas pomiaru poruszono
nadgarstkiem lub ciałem.
Urządzenie nie
wykonuje pomiaru.
Odczyty są zbyt
Pozycja mankietu jest nieprawidłowa.
wysokie lub zbyt
niskie.

Pozostałe

Wartość różni się od zmierzonej
w przychodni lub w gabinecie lekarskim.

Zalecane działanie
Wymienić wszystkie baterie na nowe.
Włożyć ponownie baterie do komory, ustawiając
ich bieguny zgodnie z informacjami w komorze.

Wymienić wszystkie baterie na nowe.

Zapiąć mankiet prawidłowo.
Podczas pomiaru należy pozostawać w bezruchu
i nie rozmawiać.
Usiąść wygodnie i w bezruchu.
Oprzeć rękę na stole z dłonią skierowaną ku
górze i mankietem założonym na wysokości
serca.
Jeżeli tętno jest bardzo słabe lub nieregularne,urządzenie może mieć problemy z określeniemciśnienia tętniczego.
Patrz część „Dlaczego warto mierzyć ciśnienietętnicze w warunkach domowych?”.
Wyjąć baterie. Następnie włożyć je prawidłowo
i ponowić pomiar.

Uwaga: jeśli czynności opisane powyżej nie rozwiążą problemu, należy skontaktować się ze sprzedawcą. Nie
próbować otwierać ani naprawiać produktu samodzielnie, gdyż spowoduje to unieważnienie gwarancji.
Polski15

Konserwacja

Nie należy otwierać urządzenia. Zawiera ono delikatne podzespoły elektryczne oraz skomplikowany systempowietrzny, które mogą łatwo ulec uszkodzeniu. Jeżeli rozwiązanie problemu na podstawie wskazówek
dotyczących rozwiązywania problemów nie powiedzie się, należy skontaktować się z autoryzowanym
dystrybutorem lub działem obsługi klienta. Dział obsługi klienta firmy A&D udzieli informacji technicznych, informacjio częściach zamiennych oraz urządzeniach autoryzowanym dystrybutorom.
Urządzenie zostało zaprojektowane i wyprodukowane tak, aby zapewnić jak najdłuższy czas eksploatacji. Mimo to,
aby zapewnić prawidłowe funkcjonowanie i dokładność pomiarów, zaleca się przeprowadzenie kontroliciśnieniomierza co 2 lata. W tym celu należy skontaktować się z najbliższym autoryzowanym dystrybutorem firmy
A&D.

Dane techniczne
Typ
Metoda pomiaru
Zakres pomiaru
Dokładność pomiaru
Zasilanie
Liczba pomiarów
Obwód nadgarstka
Klasyfikacja
Zastosowana część
Okres użytkowania

UB-525
Pomiar oscylometrycznyCiśnienie:
0–299 mmHg
Ciśnienie skurczowe:
60–279 mmHg
Ciśnienie rozkurczowe:
40–200 mmHg
Tętno:
40–180 ud./minCiśnienie: ±3 mmHg
Tętno:±5%
2 baterie alkaliczne 1,5 V (LR03 lub AAA)Ok. 250 pomiarów w przypadku używania baterii alkalicznych AAA przy ciśnieniu
wynoszącym 170 mmHg i temperaturze pokojowej 23°C.
13,5–21,5 cm
Urządzenie ME zasilane wewnętrznie (tryb pracy ciągłej)
Mankiet typu BF
Urządzenie: 5 lat (w przypadku używania sześć razy dziennie)
Polski 16

Test kliniczny

Wg ISO81060-2：2013
Podczas klinicznego badania walidacyjnego, model K5 był stosowany u 85
uczestników w celu określenia rozkurczowego ciśnienia tętniczego.
IEC 60601-1-2: 2014
Ostatnie 60 pomiarów
od +10 do +40°C/od 15 do 85% wilgotności wzgl./od 800 do 1060 hPa

EMDPamięć
Warunki pracy
Warunkiod -20 do +60°C/od 10 do 95% wilgotności wzgl./od 700 do 1060 hPatransportowe/magazynowania
WymiaryOk. 56 [szer.] x 88 [wys.] x 21,5 [gł.] mm
Masa
Ok. 90 g bez baterii
Stopień ochrony
IP20
Uwaga: ze względu na proces udoskonalania urządzeń, specyfikacja może ulec zmianie bez wcześniejszego
powiadomienia.
Klasa IP odpowiada stopniowi ochrony obudowy wg normy IEC 60529. To urządzenie jest chronione przed
ciałami stałymi o średnicy 12 mm i większej (np. palcami). Urządzenie niezabezpieczone przez działaniem
wody.

Polski17

NOTATKI

Polski18

ObsahVážený zákazníku................................................................................................................................................ 2Předběžné informace........................................................................................................................................... 2
Bezpečnostní upozornění .................................................................................................................................... 2
Identifikace součástí ............................................................................................................................................ 4
Symboly ............................................................................................................................................................... 5Používání monitoru .............................................................................................................................................. 7
1. Instalace / výměna baterií .......................................................................................................................... 7
2. Nasazení manžety ..................................................................................................................................... 8
3. Způsob přesného měření ........................................................................................................................... 8
4. Měření ........................................................................................................................................................ 9
5. Po měření ................................................................................................................................................... 9Měření................................................................................................................................................................ 10
Vyvolání dat z paměti ........................................................................................................................................ 11
Odstranění všech dat uložených v paměti ......................................................................................................... 11
Co znamená kontrolka IHB/AFib? ..................................................................................................................... 12
Co znamená AFib? ............................................................................................................................................ 12
Sloupcový ukazatel tlaku ................................................................................................................................... 12
Ukazatel klasifikace WHO ................................................................................................................................. 13
Informace o krevním tlaku ................................................................................................................................. 13
Co je krevní tlak? ......................................................................................................................................... 13
Co je to vysoký tlak a jak se léčí? ................................................................................................................ 13
Proč si měřit krevní tlak doma? .................................................................................................................... 14
Klasifikace krevního tlaku dle světové zdravotnické organizace (WHO) ...................................................... 14
Kolísání krevního tlaku ................................................................................................................................. 14
Odstraňování poruch ......................................................................................................................................... 15Údržba ............................................................................................................................................................... 16
Technické údaje................................................................................................................................................. 16Čeština1Vážený zákazníku
Blahopřejeme vám k zakoupení moderního monitoru krevního tlaku A&D. Přístroj byl navržen pro snadné použití a vysokou přesnost a
usnadňuje tak každodenní měření krevního tlaku.
Doporučujeme vám si prostudovat tuto příručku velmi pečlivě předtím, než použijete zařízení poprvé.

Předběžné informace
Toto zařízení splňuje požadavky směrnice 93/42 EHS pro lékařská zařízení. To je vyznačeno pomocí symbolu shody,referenční číslo označuje zapojení oznámeného subjektu).
Zařízení je určeno pro použití na dospělých osobách, nikoliv novorozencích nebo batolatech.
Prostředí pro použití. Zařízení je určeno k obsluze samotným pacientem v prostředí domácí zdravotní péče.
Zařízení je určeno k měřené krevního tlaku a tepové frekvence osob pro účely diagnostiky.

(0123:

Bezpečnostní upozornění
Konstrukce tohoto přístroje obsahuje přesné a citlivé součásti. Je nutné se vyhnout extrémním hodnotám teploty, vlhkosti, přímého
slunečného světla, rázům nebo prachu.
Zařízení čistěte suchým, měkkým hadrem nebo hadrem namočeným ve vodě a neutrálním čisticím prostředku. Nikdy k čištění zařízení
nepoužívejte alkohol, benzen, ředidlo nebo jiné agresivní chemikálie.
Vyhněte se těsnému skládání manžety na delší dobu, protože takové uložení může zkrátit životnost součástí.
Zařízení a manžeta nejsou voděodolné. Zabraňte pronikání deště, vody a potu do zařízení a případnému znečištění.
Měření může být zkresleno, pokud je zařízení používáno v blízkosti televizních přijímačů, mikrovlnných trub, mobilních telefonů,rentgenových nebo jiných zařízení, která generují silné elektrické pole.
Použitá zařízení, součásti a baterie nesmí být považovány za běžný domácí odpad a musí být likvidovány v souladu s platnými
místními předpisy.
Pokud zařízení používáte opakovaně, potvrďte, že je zařízení čisté.
Zařízení neměňte. Mohlo by dojít k nehodám nebo poškození zařízení.
Chcete-li měřit krevní tlak, musí být paže stlačena manžetou dostatečně natolik, aby došlo k dočasnému zastavení proudění krve
tepnou. Může to způsobit bolest, znecitlivění nebo dočasné zarudnutí zápěstí. Tento stav se objeví, obzvláště pokud je měření
opakováno několikrát po sobě. Jakákoliv bolest, znecitlivění nebo dočasné zarudnutí paže po čase zmizí.

Čeština2

Bezdrátová komunikační zařízení, například zařízení domácí sítě, mobilní telefony, bezdrátové telefony a jejich základnové stanice,krátkovlnné vysílačky atd. mohou ovlivnit funkci monitoru krevního tlaku. Proto je nutné dodržovat minimální vzdálenost 30 cm od
takových zařízení.
Příliš časté měření krevního tlaku může způsobit zranění v důsledku narušení proudění krve. Zkontrolujte, zda činnost zařízení
nezpůsobuje dlouhodobé narušení krevního oběhu, pokud zařízení používáte opakovaně.
Klinické testování nebylo provedeno na novorozencích a těhotných ženách. Nepoužívejte zařízení pro novorozence a těhotné ženy.
Pokud vám byla provedena amputace prsu, před použitím zařízení se informujte u ošetřujícího lékaře.
Zabraňte používání zařízení samotnými dětmi a nepoužívejte zařízení na místě, kde je dostupné kojencům. Mohlo by dojít k nehodám
nebo poškození.
Zařízení obsahuje malé součásti, které mohou představovat nebezpečí udušení, pokud by je kojenci omylem spolkli.
Nedotýkejte se baterií a pacienta současně. Mohlo by to způsobit úraz elektrickým proudem.
V případě poruchy jednotlivých součástí může být pouzdro blízké manžety horké a způsobit případnou poruchu.
Použití příslušenství, které není uvedeno v této příručce, může narušit bezpečnost.
Pokud by došlo ke zkratování baterie, může se velmi zahřívat a potenciálně způsobit popáleniny.
Ponechte zařízení, aby se přizpůsobilo okolnímu prostředí před používáním (asi jednu hodinu).
Nenafukujte manžetu než jejího obtočení okolo nadloktí.

Kontraindikace

Následující informace jsou bezpečnostní upozornění pro správné použití zařízení.
Nenasazujte zápěstí s připojeným jiným lékařským elektrickým zařízením. Zařízení by nemusel pracovat správně.
Osoby, které mají závažné oběhové potíže v paži, se musí informovat u svého lékaře ještě před použitím zařízení, aby se zabránilo
dalším zdravotním komplikacím.
Neprovádějte vlastní vyhodnocení výsledků měření, ani se nepokoušejte o vlastní léčbu. Vyhodnocení výsledků měření a stanovení
léčby vždy ponechte na ošetřujícím lékaři.
Nepoužívejte zařízení na zápěstí s nezahojeným zraněním.
Nenasazujte manžetu na paži s připojenou kanylou intravenózní infuze nebo krevní transfuze. Mohlo by dojít ke zranění nebo
nehodám.
Nepoužívejte zařízení tam, kde se vyskytují hořlavé plyny, například anestetické plyny. Mohlo by dojít k výbuchu.
Nepoužívejte zařízení v místech s vysokou koncentrací kyslíku, například ve vysokotlaké kyslíkové komoře nebo v kyslíkovém stanu.
Mohlo by dojít k požáru nebo výbuchu.

Čeština3

Identifikace součástí
DisplejTlačítko(průměr, paměť a změna nastavených parametrů)Tlačítko START
Kryt baterieManžetaPAMĚŤPrůměr

Displej

Alkalické baterie (LR03 nebo AAA)

Systolický tlak
Diastolický tlak

Ukazatel klasifikace WHOaSloupcový ukazatel tlaku

Symbol IHB/AFib

Symbol srdce

Tepová frekvence

Kontrolka baterieČeština4Symboly

Symboly, které jsou umístěny na pouzdře zařízení
Symboly

Funkce / význam
Pohotovostní režim a zapnutí zařízení

SYS

Systolický krevní tlak, mmHg

DIA

Diastolický krevní tlak, mmHg

PUL

Tepy za minutuPrůvodce instalací baterie
Stejnosměrný proud

SN

Sériové číslo
Datum výroby
Typ BF: zařízení a manžeta jsou navrženy tak, aby zajišťovaly speciální ochranu proti úrazu
elektrickým proudem.
Směrnice EU – štítek lékařského zařízení

IP

Mezinárodní symbol ochranyŠtítek WEEEVýrobce
Zástupce EU
Viz uživatelská příručka / brožura
Udržujte v suchuČeština5Symboly, které se objevují na displeji
Symboly

Funkce / Význam / Doporučené opatření
Zobrazuje se v průběhu měření. Bliká, pokud je detekován tep.
Zůstaňte v klidu.
Symbol IHB/AFib se zobrazuje v okamžiku, kdy je detekován nepravidelný srdeční tep.
Může svítit, když je detekována velmi jemná vibrace, například chvění nebo třes.

Symboly, které se objevují na displeji (pokračování)Symboly

Funkce / význam
Předchozí měření uložené v PAMĚTI.
Průměrná data.
ZCELA NABITÁ BATERIE
Kontrolka nabití baterie během měření.
VYBITÁ BATERIE
Pokud kontrolka bliká, je baterie téměř vybitá.

Nestabilní krevní tlak v důsledku pohybu během měření.
nebo

Doporučené opatření

Pokud kontrolka bliká, vyměňte
všechny baterie za nové.
Proveďte další měření.
Během měření zůstaňte nehybní.

Hodnoty systolického a diastolického tlaku jsou vzájemně v
rozsahu 10 mmHg.
Hodnota tlaku se během nafukování nezvyšuje.
Manžeta není nasazena správně.
CHYBA ZOBRAZENÍ TEPU
Tep není detekován správně.

Nasaďte správně manžetu a opakujteměření.

Vnitřní chyba monitor krevního tlaku.

Vyjměte baterie a stiskněte tlačítko
START , pak baterie vložte zpět.
Pokud se chyba stále projevuje,
kontaktujte prodejce.
Čeština6

1. Instalace / výměna bateriíPoužívání monitoru

1. Sejměte kryt baterie.
2. Vyjměte použité baterie a vložte nové do oddílu pro baterie,
jak je znázorněno na obrázku. Zajistěte přitom správnou
orientaci s ohledem na polaritu (+ a -) baterií. Používejte
pouze baterie LR03 nebo AAA.

Kryt baterie

3. Upevněte kryt baterie.

Použité baterieUPOZORNĚNÍ
Nové baterieVložte baterie způsobem znázorněným na obrázku do oddílu
pro baterie.
Pokud je instalace nesprávná, zařízení nebude funkční.
Když se na displeji rozbliká kontrolka(symbol VYBITÁBATERIE), vyměňte baterie za nové. Nemíchejte staré a novébaterie. Mohlo by dojít ke zkrácení životnosti baterií nebo kporuše zařízení.
Kontrolka (symbol VYBITÁ BATERIE) se nezobrazuje,když jsou baterie vybité.
Životnost baterií se liší podle okolní teploty a při nízkých teplotách se může zkracovat. Obecně platí, že dvě novébaterie LR03/AAA vydrží alespoň čtyři měsíce, pokud zařízení bude používáno ke dvěma měřením denně.
Používejte pouze určené baterie. Baterie dodané se zařízením jsou určeny pro testování výkonnosti monitoru amohou mít omezenou životnost.
Vyjměte baterie, pokud jste zařízení dlouho nepoužívali. Z baterií může unikat elektrolyt a způsobit poruchu.

Čeština7

2. Nasazení manžety
1. Obtočte manžetu okolo zápěstí asi 1 cm nad dlaní, jak je znázorněno na obrázku
vpravo.

Asi 1 cm

2. Nasazení manžety pomocí suchého zipu.
Poznámka:

Pro přesná měření upevněte manžetu těsně a měřte na holém zápěstí.

3. Způsob přesného měření

Pro nejpřesnější měření krevního tlaku:Během měření se nehýbejte a buďte zticha.
Pohodlně se posaďte. Položte loket a ruce dlaněmi nahoru na stůl a manžetu nasaďte do stejné výšky jako je
srdce.
Před měřením se asi na pět až deset minut uvolněte. Pokud pociťujete emocionální vzrušení nebo naopak depresi,měření bude tento stres odrážet ve formě vyššího (nebo nižšího) krevního tlaku, než je normální hodnota, a tepová
frekvence bude obvykle vyšší než normální hodnota.
Pokoušejte se měřit krevní tlak ve stejný čas každý den.
Tlak krve se velmi liší, v závislosti na tom, co děláte, co jste jedli a pili - má to velmi silný a rychlý vliv na váš krevní
tlak.
Neprovádějte měření ihned po fyzické námaze nebo po koupání. Před měřením buďte v klidu alespoň dvacet nebotřicet minut.
Nezkřižujte nohy. Udržujte nohy naplocho na podlaze a narovnejte si záda.
Toto zařízení zakládá své měření na tepové frekvenci. Pokud máte velmi slabou nebo nepravidelnou tepovou
frekvenci, zařízení může mít obtíže se stanovením krevního tlaku.
Pokud by zařízení detekovalo stav, který je neobvyklý, zastaví měření a zobrazí symbol chyby. Popis symbolů viz
strana 6.
Tento monitor krevního tlaku je určen pro používání dospělými osobami. Před použitím zařízení na měření u dítěte
se informujte u ošetřujícího lékaře. Dítě by nemělo toto zařízení používat bez dozoru.
Automatická funkce monitoru krevního tlaku může být ovlivněna nadměrnou teplotou, vlhkostí nebo nadmořskou
výškou.
Čeština8

4. Měření

Během měření je normální, pokud se manžeta velmi těsně stáhne.

5. Po měření

Po zobrazení odečtů měření, pokud stiskněte tlačítko START a vypněte zařízení, jsou nové odečty uloženy dopaměti.
Po zobrazení odečtů měření, pokud stiskněte tlačítko
a vypněte zařízení, nové odečty nejsou uloženy dopaměti.
Sejměte manžetu a zaznamenejte data.
Poznámka: Zařízení má funkci automatického vypnutí napájení, která uloží aktuální data do paměti a zařízení
vypne asi po jedné minutě od ukončení měření. Ponechte alespoň tři minuty mezi jednotlivými
měřeními u jedné osoby.

Čeština9MěřeníPoznámka: UB-525 při použití poskytne nafukování vhodné pro daného uživatele.
1. Obtočte manžetu okolo zápěstí. Umístěte střed manžety do stejné úrovně sesrdcem a uvolněte se.
2. Stiskněte tlačítko START . Všechny segmenty displeje jsou zobrazeny.
3. Zobrazí se nula (0) a krátce problikne. Jakmile začne měření, displej se změní.
Manžeta se začne nafukovat. Je normální, pokud se manžeta velmi těsně
stáhne. Jakmile je nafukování dokončeno, automaticky se spustí měření arozbliká se symbol(srdce).
Poznámka: Pokud chcete kdykoliv zastavit nafukování, stiskněte tlačítko
START ještě jednou.
4. Když je měření dokončeno, zobrazí se odečty (systolického a diastolického
tlaku a tepové frekvence, klasifikace WHO, symbol I.H.B). Manžeta vypustízbývající tlak a automaticky se úplně vyfoukne.
Poznámka:

Pokud nechcete nové měření uložit do paměti, stiskněte
tlačítkopři zobrazení naměřených hodnot.

5. Stiskněte znovu tlačítko START a vypněte zařízení. Sejměte manžetu.
Poznámka: Zařízení má funkci automatického vypnutí napájení.
Ponechte alespoň tři minuty mezi jednotlivými měřeními u jedné
osoby.

Čeština10

Všechny segmenty
Zobrazení nuly
Nafukování jespuštěno
Probíhá nafukování aměření
Systolický tlak
Diastolický tlak
Klasifikace WHO
Symbol I.H.B.
Tepová frekvenceVyvolání dat z paměti
Poznámka: Zařízení dokáže do paměti uložit 60 posledních měření.
a vypněte zařízení.
1. Stiskněte tlačítko
Zobrazí se průměr ze všech měření a počet naměřených dat. Pokudžádná data nejsou naměřena, zobrazí se „0“. Stiskněte tlačítko
nebo
START a vypněte zařízení.
2. Pomocí následujících tlačítek zobrazte data (počet a naměřená data).
Zařízení zobrazí průměr ze všech měření a počet naměřených dat.
Při každém stisknutí tlačítka
zařízení zobrazí následující údaje:Průměrná data ze všech provedených měření.
Data (počet a naměřená data).
Zařízení zobrazí údaje v pořadí od nejnovějších dat.
Příklad: č.30 a data
č.29 a datač.01 a data.
po zobrazení nejstarších dat, zařízení
3. Pokud stisknete tlačítko
pokračuje ke kroku 1 a zobrazí průměr ze všech měření a početnaměřených dat.
4. Stiskněte znovu tlačítko START a vypněte zařízení.
Po jedné minutě nečinnosti se zařízení automaticky vypne.

Režim paměti

tlačítkoPrůměr ze všech dat
Počet měření dat
tlačítkoPříklad: č. 30
Nejnovější datatlačítko
Nejstarší data
(poslední)Odstranění všech dat uložených v paměti
1. Stiskněte a podržte tlačítko

, dokud se zařízení automaticky nevypne.

Zařízení zobrazí symboly, odstraní data uložená v paměti a současně se
rozbliká symbol, pak se zařízení automaticky vypne.
Čeština11

Vypnouttlačítko

VypnouttlačítkoCo znamená kontrolka IHB/AFib?
Když monitor detekuje nepravidelný rytmus během měření, kontrolka IHB/AFib se zobrazuje na displeji společně snaměřenými hodnotami.
Poznámka: Doporučujeme kontaktovat vašeho ošetřujícího lékaře, pokud se kontrolka
zobrazovat často.

IHB/AFib bude

Co znamená AFib?

Srdce se stahuje v důsledku elektrických signálů, které vznikají v srdci, a čerpá krev oběhovým systémem. Fibrilace
síní (AFib) nastává, když je elektrický signál v srdeční síni chybný a vede k narušení intervalu srdečního tepového
impulsu. Fibrilace síní (AFib) může způsobit, že krev stagnuje v srdci, což může snadno vytvořit krevní sraženiny av důsledku toho mrtvici nebo infarkt.
Tlakování

Sloupcový ukazatel tlaku

Ukazatel monitorujte průběh tlaku během měření.
Probíhá nafukování
Probíhá měřeníNafukování dokončenoČeština12

Ukazatel klasifikace WHOKaždý ze šesti segmentů sloupcového ukazatele odpovídá
klasifikaci krevního tlaku dle organizace WHO a je popsán na
straně 14.
PříkladStředně zvýšený Málo zvýšený tlakMírně zvýšený
tlak
normální tlak

Ukazatel klasifikace WHO
Velmi vysoký tlakStředně zvýšený tlak
Málo zvýšený tlakMírně zvýšený normální tlak
Normální
Optimální tlak

: Ukazatel na základě aktuálních dat

zobrazuje segment, který odpovídá klasifikaceorganizace WHO.

Co je krevní tlak?

Informace o krevním tlaku

Krevní tlak je mírou síly, kterou působí krev na stěny tepen. Systolický tlak nastává, když se srdce stahuje.
Diastolický tlak nastává, když se srdce roztahuje. Krevní tlak se měří v milimetrech rtuťového sloupce (mmHg).
Přirozený krevní tlak je vyjadřován základním tlakem, který je měřen ihned po ránu, když je měřená osoba v klidu ana lačno.

Co je to vysoký tlak a jak se léčí?

Vysoký tlak, hypertenze, představuje neobvykle vysoký krevní tlak, který v případě neléčení může způsobit mnoho
problémů, včetně mrtvice nebo infarktu. Vysoký tlak lze léčit změnou životního stylu, vyhýbáním se stresu amedikací pod dohledem lékaře.
Zabránění vysokému tlaku nebo jeho udržení pod kontrolou:NekuřtePravidelně cvičteSnižte objem přijímané soli a tuku
Nechte se pravidelně fyzicky vyšetřovat
Udržujte si správnou hmotnostČeština13

Proč si měřit krevní tlak doma?

Krevní tlak měřený na klinice nebo v ordinaci lékaře může způsobit obavu a způsobit chybně vyšší měření, o 25 až
30 mmHg vyšší než při měření doma. Měření doma snižuje dopad vnějších vlivů na odečty krevního tlaku, doplňuje
měření lékařem a poskytuje přesnější a úplnější historii měření krevního tlaku.
Referenční materiál: Journal of

Klasifikace krevního tlaku dle světové zdravotnické
organizace (WHO)Kolísání krevního tlaku

Krevní tlak jednotlivce velmi kolísá během dne a podle ročního
období. Může se lišit o 30 až 50 mmHg v důsledku různých
podmínek během dne. U osob s vysokým krevním tlakem je kolísání
ještě výraznější. Normálně se krevní tlak zvyšuje v práci neboběhem hry a klesá na nejnižší hodnoty během spánku. Nebuďte
proto příliš znepokojeni výsledky jednoho měření.
Měření provádějte ve stejný čas dne, každý den, podle postupupředepsaného v této příručce, abyste zjistili svůj normální krevní tlak.
Pravidelné odečty poskytují komplexnější historii krevního tlaku.
Nezapomeňte si poznamenat datum a čas měření krevního tlaku.
Informujte se u svého lékaře a interpretujte data měření krevního
tlaku.

Vysoký tlak stupně 3 (velmi vysoký)Vysoký tlak stupně 2 (středně zvýšený)Vysoký tlak stupně 1 (málo zvýšený)Mírně zvýšený normální tlak

Normální
Optimální tlak

Systolický krevní tlak

Typické kolísání během dne(měřeno během každých pěti minut)
Systolický
Krevní tlakSvětovou zdravotnickou organizací (WHO) byly stanoveny
standardy pro hodnocení vysokého krevního tlaku bez ohledu
na věk, jak je znázorněno v tabulce vpravo.

Diastolický krevní tlak

Hypertension 1999, svazek 17, č. 2

Diastolický
OdpoledneČeština14

SpánekDopoledneČasOdstraňování poruch
ProblémNa displeji se nic
nezobrazuje, ani kdyžje zařízení zapnuto.

Manžeta se
nenafukuje.

Zařízení neměří.
Odečty jsou příliš
vysoké nebo příliš
nízké.

Jiné
Poznámka:Možná příčina
Baterie jsou vybité.

Doporučené opatření
Vyměňte baterie za nové.
Vložte baterie znovu se záporným a kladným
Póly baterie nejsou správně orientovány. pólem tak, aby to odpovídalo označení v oddílu
pro baterie.
Napětí baterií je nízké.
Bliká kontrolka (symbol VYBITÁ
Vyměňte baterie za nové.
BATERIE).
Pokud jsou baterie úplně vybité, symbol
se nezobrazí.
Manžeta není nasazena správně.
Nasaďte manžetu správně.
Během měření jste přesunuli zápěstí
Zajistěte, abyste během měření seděli v klidu anebo tělo.
tiše.
Usaďte se pohodlně a v klidu.
Poloha manžety je nesprávná.
Položte ruce na stůl dlaněmi nahoru a manžetou
nasaďte do stejné výšky jako je srdce.
Pokud máte velmi slabou nebo nepravidelnou
tepovou frekvenci, zařízení může mít obtíže sestanovením krevního tlaku.
Hodnota se odlišuje od hodnoty
Viz část „Proč si měřit krevní tlak doma?“.
naměřené lékařem nebo na klinice.
Vyjměte baterie. Umístěte je správně zpět aopakujte měření.

Pokud výše uvedené akce nevyřeší problém, kontaktujte prodejce. Neotevírejte ani neopravujte tento
produkt, protože by to způsobilo neplatnost poskytované záruky.
Čeština15ÚdržbaZařízení neotevírejte. Používá citlivé elektrické součásti a jemnou pneumatickou jednotku, které lze snadno
poškodit. Pokud problém nedokážete odstranit podle pokynů pro odstraňování poruch, kontaktujte autorizovaného
prodejce ve svém místě nebo naše oddělení péče o zákazníky. Oddělení péče o zákazníky společnosti A&D
poskytne technické informace, náhradní součásti a jednotky autorizovanému prodejci.
Zařízení bylo zkonstruováno a vyrobeno tak, aby dosahovalo dlouhé provozní životnosti. Obecně se však
doporučuje, nechat zařízení zkontrolovat každé 2 roky a zajistit tak správnou funkci a přesnost. Kontaktujte
autorizovaného prodejce nebo společnost A&D, pokud potřebujete provést údržbu zařízení.

Technické údaje
TypZpůsob měření
Rozsah měření

Přesnost měření
Napájecí zdroj
Počet měření
Obvod zápěstí
KlasifikacePoužitá částUžitečná životnost

UB-525
Oscilometrické měření
Tlak:
0–299 mmHg
Systolický tlak:
60–279 mmHg
Diastolický tlak:
40–200 mmHg
Tepová frekvence: 40–180 tepů / minuta
Tlak:±3 mmHg
Tepová frekvence: ±5 %2 x alkalické baterie 1,5 V (LR03 nebo AAA)Přibližně 250 měření při použití alkalických baterií AAA, při hodnotě tlaku 170 mmHg
a pokojové teplotě 23 °C.
13,5 – 21,5 cm
Vnitřní napájení zařízení (ME) (režim nepřetržitého provozu)
Manžeta typ BF
Zařízení: 5 let (při použití šestkrát denně)Čeština16

Klinický test
EMD
Paměť
Provozní podmínkyPřepravní / skladovací
podmínkyRozměry
Hmotnost
Ochrana proti vniknutí

Podle normy ISO 81060-2 : 2013
V klinické validační studii používalo přístroj K5 85 subjektů ke stanovení diastolického
krevního tlaku.
IEC 60601-1-2: 2014
Posledních 60 měření
+10 až +40 °C / relativní vlhkost 15 až 85 % / 800 až 1 060 hPa-20 až +60 °C / relativní vlhkost 10 až 95 % / 700 až 1 060 hPaPřibližně 56 [š] x 88 [v] x 21,5 [h] mmPřibližně 90 g, bez baterií
IP20

Poznámka: Technické údaje se mohou při vylepšování měnit bez předchozího upozornění.
Klasifikace IP představuje stupeň ochrany zajištěné skříní v souladu s IEC 60529. Zařízení je chráněno
před pronikáním pevných cizích těles o průměru 12 mm a větším, například prstů. Zařízení neníchráněno před průnikem vody.

Čeština17

POZNÁMKY

Čeština18

ObsahVážení zákazníci.................................................................................................................................................. 2Úvodné poznámky ............................................................................................................................................... 2
Opatrenia ............................................................................................................................................................. 2
Identifikácia dielov ............................................................................................................................................... 4
Symboly ............................................................................................................................................................... 5Používanie monitora ............................................................................................................................................ 7
1. Inštalácia/výmena batérií ........................................................................................................................... 7
2. Nasadenie manžety ................................................................................................................................... 8
3. Postup presného merania .......................................................................................................................... 8
4. Meranie ...................................................................................................................................................... 9
5. Po meraní ................................................................................................................................................... 9
Merania .............................................................................................................................................................. 10
Načítanie údajov v pamäti ................................................................................................................................. 11
Vymazanie údajov uložených v pamäti .............................................................................................................. 11Čo je indikátor IHB/AFib? .................................................................................................................................. 12Čo je AFib? ........................................................................................................................................................ 12
Paličkový indikátor tlaku .................................................................................................................................... 12
Indikátor klasifikácie SZO .................................................................................................................................. 13
O krvnom tlaku................................................................................................................................................... 13Čo je krvný tlak? ........................................................................................................................................... 13Čo je hypertenzia a ako sa upravuje? .......................................................................................................... 13
Prečo merať krvný tlak doma? ..................................................................................................................... 14
Klasifikácia SZO krvného tlaku..................................................................................................................... 14Odchýlky krvného tlaku ................................................................................................................................ 14
Riešenie problémov ........................................................................................................................................... 15Údržba ............................................................................................................................................................... 16
Technické údaje................................................................................................................................................. 16

Slovenčina

1Vážení zákazníci
Blahoželáme vám k zakúpeniu špičkového tlakomera spoločnosti A&D. Toto zariadenie, ktoré je navrhnuté tak, aby
sa ľahko používalo a bolo presné, vám uľahčí každodenné meranie krvného tlaku.
Odporúčame, aby ste si túto príručku pozorne prečítali pred prvým použitím zariadenia.

Úvodné poznámky
Toto zariadenie je v súlade s európskou smernicou 93/42/EHS o zdravotníckych pomôckach. Túto skutočnosť
označuje značka zhody. (0123: Referenčné číslo pre príslušný notifikovaný orgán).
Zariadenie je určené na meranie tlaku dospelých, nie novorodencov alebo dojčiat.
Prostredie na používanie zariadenia. Zariadenie je určené na prevádzku používateľmi v prostredí domácej
zdravotnej starostlivosti.
Toto zariadenie je určené na meranie krvného tlaku a pulzu ľudí na účely určenia diagnózy.

Opatrenia
V tomto zariadení sú použité presné súčasti. Je potrebné predchádzať extrémnym teplotám, vlhkosti, priamemu
slnečnému žiareniu, nárazom a prachu.
Zariadenie čistite suchou a mäkkou látkou alebo látkou namočenou vo vode a neutrálnym čistiacim prostriedkom.
Na čistenie zariadenia nikdy nepoužívajte alkohol, benzén, rozpúšťadlo alebo iné silné chemikálie.
Manžetu neskladajte napevno a dlhodobo, pretože to môže skrátiť životnosť týchto komponentov.
Zariadenie nie je odolné voči vode. Zabráňte styku dažďa, potu alebo vody so zariadením.
Merania môžu byť skreslené, ak sa zariadenie používa v blízkosti televízorov, mikrovlnných rúr, mobilných
telefónov, röntgenových alebo iných zariadení so silnými elektrickými poľami.
Použité zariadenie, jeho diely a batérie sa nepovažujú za bežný komunálny odpad, a preto sa musia likvidovať
podľa príslušných miestnych predpisov.
Pri opätovnom použití zariadenia skontrolujte, či je čisté.
Zariadenie neupravujte. Mohlo by dôjsť k úrazu alebo poškodeniu zariadenia.
Ak chcete merať krvný tlak, manžeta musí dostatočne silno stlačiť zápästie, aby sa dočasne zastavil prietok krvi cez
tepnu. To môže spôsobiť bolesť, znecitlivenie alebo dočasný červený otlačok na zápästí. Tieto stavy sa objavia
najmä vtedy, ak sa meranie opakuje. Akákoľvek bolesť, znecitlivenie alebo červené otlačky časom zmiznú.
Bezdrôtové komunikačné zariadenia, ako sú domáce sieťové zariadenia, mobilné telefóny, bezdrôtové telefóny, ich
Slovenčina 2

základňové stanice a vysielačky môžu ovplyvniť tento tlakomer. Preto by sa mala dodržiavať minimálna vzdialenosť
30 cm od týchto zariadení.
Pričasté meranie krvného tlaku môže poškodiť ujmu z dôvodu rušenia prietoku krvi. Skontrolujte, či prevádzka
zariadenia nemá za následok dlhodobé zhoršenie krvného obehu pri opakovanom použití zariadenia.
Klinické skúšky sa nevykonávali na novorodencoch a tehotných ženách. Zariadenie nepoužívajte v prípade
novorodencov alebo tehotných žien.
Ak ste podstúpili mastektómiu, pred použitím zariadenia sa poraďte sa s lekárom.
Nedovoľte deťom samostatne používať zariadenie a nepoužívajte zariadenie na mieste v dosahu dojčiat. Mohlo bydôjsť k úrazu alebo poškodeniu.
Zariadenie obsahuje malé diely, ktoré môžu spôsobiť nebezpečenstvo udusenia, ak ich dojča omylom prehltne.
Nedotýkajte sa súčasne batérií a pacienta. Môže to spôsobiť úraz elektrickým prúdom.
V prípade zlyhania jednotlivých komponentov okolie v blízkosti manžety môže byť horúce a môže spôsobiť poruchu.
Používanie príslušenstva, ktoré nie je uvedené v tejto príručke, môže ohroziť bezpečnosť.
Ak nastane skrat batérie, môže byť horúca a môže spôsobiť popáleniny.
Zariadenie pred použitím nechajte, aby sa adaptovalo na okolité prostredie (približne jednu hodinu).
Manžetu nenafukujte, keď nie je omotaná okolo zápästia.

KontraindikácieNižšie sú uvedené opatrenia na správne používanie zariadenia.
Manžetu nenasadzujte na zápästie s iným pripojeným elektrickým zdravotným zariadením. Zariadenie by nemuselo
fungovať správne.
Ľudia, ktorí majú ťažký deficit obehu v ramene, sa musia pred použitím zariadenia poradiť s lekárom, aby predišlizdravotným problémom.
Nestanovujte vlastnú diagnózu na základe výsledkov meraní a sami nezačínajte liečbu. Vždy sa o vyhodnotenívýsledkov a liečbe poraďte so svojím lekárom.
Zariadenie nenasadzujte na zápästie s nevyliečenou ranou.
Zariadenie nenasadzujte na rameno s intravenóznou infúziou alebo krvnou transfúziou. Mohlo by dôjsť k zraneniu
alebo poškodeniu.
Zariadenia nepoužívajte v prítomnosti horľavých plynov, ako sú napríklad anestetické plyny. Mohlo by dôjsť k explózii.
Zariadenie nepoužívajte v prostredí s vysokou koncentráciou kyslíka, ako napríklad vysokotlaková kyslíková
komora alebo kyslíkový stan. Mohlo by dôjsť k požiaru alebo výbuchu.
Slovenčina

3

Identifikácia dielov
Displej

Tlačidlo(priemer, pamäť a zmena vopred nastavených parametrov)

Tlačidlo START
Kryt batériíManžetaPAMÄŤ
Priemer

Alkalické batérie (LR03 alebo AAA)

Displej
Systolický tlak
Diastolický tlak

Indikátor klasifikácie SZOaPaličkový indikátor tlaku

Symbol IHB/AFib

Symbol srdca

Pulz

Indikátor batérií
Slovenčina

4

Symboly

Symboly, ktoré sú vytlačené na kryte zariadenia
Symboly

Funkcia/Význam
Pohotovostný režim a zapnutie zariadenia

SYS

Systolický krvný tlak v mmHg

DIA

Diastolický krvný tlak v mmHg

PUL

Pulz za minútu
Návod na inštaláciu batérií
Jednosmerný prúd

SN

Sériové číslo
Dátum výroby
Typ BF: Zariadenie a manžeta sú navrhnuté tak, aby poskytovali osobitnú ochranu proti úrazom
elektrickým prúdomŠtítok zdravotníckej pomôcky podľa smernice ES

IP

Symbol medzinárodnej ochranyŠtítok WEEEVýrobca
Zástupca EÚPozrite si návod na obsluhu/brožúru
Uchovávajte v suchu
Slovenčina

5

Symboly, ktoré sa zobrazujú na displeji
Symboly

Funkcia/Význam/Odporúčaná akcia
Zobrazí sa v priebehu merania.Bliká, keď sa zistí pulz.
Hýbte sa čo najmenej.
Symbol IHB/AFib Zobrazí sa, keď sa zistí nepravidelný pulz srdca.
Môže sa rozsvietiť, keď sa zistí veľmi malá vibrácia ako chvenie alebo trasenie.

Symboly, ktoré sa zobrazujú na displeji (pokračovanie)
Symboly

Funkcia/Význam
Predchádzajúce merania uložené v PAMÄTI
Priemerné údajeNABITÁ BATÉRIA
Indikátor napájania z batérie počas merania
TAKMER VYBITÁ BATÉRIA
Batéria je takmer vybitá, keď bliká

Nestabilný krvný tlak v dôsledku pohybu počas merania
alebo

Odporúčaná akcia

Keď indikátor bliká, vymeňte všetky
batérie za nové.
Vykonajte ďalšie meranie.
Počas merania sa nehýbte.

Rozsah medzi systolickými a diastolickými hodnotami je
10 mmHg
Hodnota tlaku sa pri nafukovaní nezvýšila
Manžeta nie je nasadená správne.
CHYBA ZOBRAZENIA PUL
Pulz sa nezisťuje správneManžetu nasaďte správne a vykonajteďalšie meranie.

Vnútorná chyba tlakomera

Vyberte batérie a stlačte tlačidlo
START a potom znova nainštalujte
batérie. Ak sa chyba stále zobrazuje,obráťte sa na predajcu.
Slovenčina

6

1. Inštalácia/výmena batériíPoužívanie monitora

1. Odstráňte kryt batérií.
2. Vyberte použité batérie a do priestoru na batérie vložte nové
batérie tak, ako je znázornené, pričom sa ubezpečte, či jepolarita (+ a -) správna. Používajte iba batérie LR03 alebo AAA.

Kryt batérií

3. Nasaďte kryt batérie.
Použité batérie
UPOZORNENIA
Batérie vkladajte tak, ako je znázornené v priestore na batérie.
Nové batérie
Ak budú vložené nesprávne, zariadenie nebude fungovať.
Ak na displeji bliká symbol(TAKMER VYBITÁBATÉRIA), vymeňte všetky batérie za nové. Nemiešajte staré
a nové batérie.Mohlo by to skrátiť životnosť batérií alebo
spôsobiť poruchu zariadenia.
Symbol (TAKMER VYBITÁ BATÉRIA) sa nezobrazuje,
keď sú batérie vybité.
Životnosť batérií sa mení v závislosti od okolitej teploty a pri nízkych teplotách môže byť kratšia. Dve nové batérie
LR03/AAA vo všeobecnosti vydržia približne štyri mesiace, ak sa zariadenie používa na dve merania denne.
Používajte iba uvedené batérie. Batérie dodávané so zariadením sú určené na testovanie výkonu tlakomera amôžu mať obmedzenú životnosť.
Ak sa zariadenie nebude dlhší čas používať, vyberte batérie. Batérie by mohli vytiecť a spôsobiť poruchu.

Slovenčina

7

2. Nasadenie manžety
1. Oviňte manžetu okolo zápästia asi 1 cm nad rukou tak, ako je to znázornené naobrázku vpravo.

Asi 1 cm

2. Pomocou suchého zipsu pevne utiahnite manžetu.
Poznámka:

Ak chcete krvný tlak presne odmerať, manžetu pevne utiahnite avykonajte meranie na holom zápästí.

3. Postup presného merania

Aby ste krvný tlak odmerali čo najpresnejšie:
Počas merania zostaňte v pokoji a nerozprávajte.
Usaďte sa v pohodlnej polohe. Položte lakeť na stôl dlaňou nahor a manžetou na rovnakej úrovni ako vaše srdce.
Pred meraním sa na päť až desať minút uvoľnite. Ak ste vzrušení alebo ste v depresii v dôsledku emočného stresu,
pri meraní sa tento stres odzrkadlí ako krvný tlak, ktorý je vyšší (alebo nižší) ako normálny, a nameraný pulz bude
zvyčajne rýchlejší ako normálny.
Pokúste sa merať svoj krvný tlak každý deň približne v rovnaký čas.
Krvný tlak človeka sa neustále mení v závislosti od toho, čo robí, čo jedol a čo pil, čo môže mať veľmi silný a rýchly
vplyv na váš krvný tlak.
Meranie nevykonávajte hneď po fyzickej námahe alebo kúpeli. Pred meraním si dvadsať alebo tridsať minút
odpočiňte.
Neprekrižujte nohy. Celé chodidlá majte na podlahe a narovnajte chrbát.
Toto zariadenie meria na základe pulzu srdca. Ak máte veľmi slabý alebo nepravidelný pulz srdca, zariadenie môžemať ťažkosti s určením krvného tlaku.
Ak zariadenie zistí stav, ktorý je abnormálny, ukončí meranie a zobrazí symbol chyby. Popis symbolov si pozrite na
strane 6.
Tento tlakomer je určený pre dospelých. Pred použitím tohto zariadenia na meranie tlaku dieťaťa sa poraďte slekárom. Dieťa by nemalo toto zariadenie používať bez dozoru.
Výkonnosť automatického tlakomera môžu ovplyvniť príliš vysoká teplota, vlhkosť alebo nadmorská výška.
Slovenčina

8

4. Meranie

V priebehu merania je normálne, že sa manžeta zdá byť veľmi tesná.

5. Po meraníAk počas zobrazovania nameraných hodnôt vypnete zariadenie stlačením tlačidla START , nové namerané
hodnoty sa uložia do pamäti.
Ak počas zobrazovania nameraných hodnôt vypnete zariadenie stlačením tlačidla , nové namerané hodnoty sa
do pamäti neuložia.
Odstráňte manžetu a zaznamenajte údaje.
Poznámka: Zariadenie je vybavené funkciou automatického vypnutia napájania, ktorá uloží aktuálne údaje avypne zariadenie približne jednu minútu po meraní. Medzi meraniami tlaku rovnakej osoby
zachovávajte minimálne trojminútové prestávky.

Slovenčina

9Merania
Poznámka:

Ak použijete model UB-525, dosiahnete tak nafukovanie, ktoré je prepoužívateľa najvhodnejšie.

1. Oviňte si manžetu okolo zápästia. Pohodlne sa usaďte, manžetu umiestnite na
rovnakú úroveň, ako je vaše srdce, a uvoľnite sa.
2. Stlačte tlačidlo START . Zobrazia sa všetky segmenty displeja.

Všetky segmenty

3. Nakrátko sa zobrazí blikajúca nula (0). Keď sa začne meranie, zobrazenie na
displeji sa zmení. Manžeta sa začne nafukovať. Je normálne, že sa manžeta
zdá byť veľmi tesná. Keď sa nafukovanie spustí, automaticky sa začne meranie
a bude blikať symbol(srdce).
Poznámka:

Ak chcete kedykoľvek zastaviť nafukovanie, opätovne stlačte
tlačidlo START .

4. Keď sa meranie dokončí, na zariadení sa zobrazia namerané hodnoty(namerané hodnoty systolického a diastolického tlaku, pulzu, klasifikácie SZO,
symbolu I.H.B). Manžeta vypustí zvyšný vzduch a automaticky sa úplne
vyfúkne.
Poznámka:

Zobrazí sa nula
Nafukovanie sazačaloPrebieha nafukovanie
a meranie

Ak do pamäti nechcete ukladať nové namerané hodnoty, počas
zobrazovania nameraných stlačte tlačidlo .

5. Vypnite zariadenie opätovným stlačením tlačidla START . Odstráňte
manžetu.
Poznámky:

Zariadenie je vybavené funkciou automatického vypnutia napájania.
Medzi meraniami tlaku rovnakej osoby zachovávajte minimálne
trojminútové prestávky.

Slovenčina

10

Systolický tlak
Diastolický tlak
Klasifikácia SZOSymbol I.H.B
Pulz

Načítanie údajov v pamäti
Poznámka: Toto zariadenie ukladá do pamäte posledných 60 meraní.
1. Ak chcete zariadenie vypnúť, stlačte tlačidlo .
Zobrazí sa priemer všetkých meraní a počet údajov. Ak nie sú k dispozíciižiadne údaje, zobrazí sa „0“. Stlačením tlačidla
alebo START
vypnite zariadenie.
2. Na zobrazenie údajov (o počte a meraniach) použite nasledujúce tlačidlá.
Na zariadení sa zobrazí priemer všetkých meraní a následne aj
počet údajov.
Každým stlačením tlačidla
sa v zariadení zobrazia nasledujúceúdaje:
Priemerné údaje všetkých medzitým vykonaných meraní.
Údaje (údaje o počte a meraniach).
Zobrazia sa zariadení v poradí od najnovších údajov.
Napríklad: Č. 30 a údaje
Č. 29 a údaje
Č. 01 a údaje.
3. Ak po zobrazení najstarších údajov stlačíte tlačidlo , zariadenie
prejde na krok 1 a na zariadení sa zobrazí priemer všetkých meraní anásledne aj počet údajov.
4. Vypnite zariadenie stlačením tlačidla START .
Po jednej minúte nečinnosti sa zariadenie automaticky vypne.

Režim pamäti

tlačidlo
Priemer všetkých
údajov
Počet všetkých údajov, kým sa zariadenie automaticky nevypne.

Na zariadení sa zobrazí symbol, a kým bude symbolúdaje uložené v pamäti a zariadenie sa automaticky vypne.
Slovenčina

blikať, vymažú sa
11

tlačidlo

Napr.: č. 30
Najnovšie údaje
tlačidlo
Posledné údaje
(najstaršie)

Vymazanie údajov uložených v pamäti
1. Stlačte a podržte tlačidlo

Vypnutietlačidlo

VypnutietlačidloČo je indikátor IHB/AFib?
Ak monitor zistí počas merania nepravidelný rytmus, na displeji sa zobrazí indikátor IHB/AFib s nameranými
hodnotami.
Poznámka: Ak sa tento indikátor IHB/AFibzobrazuje často, odporúčame, aby ste kontaktovali lekára.

Čo je AFib?

Srdce sa sťahuje v dôsledku elektrických signálov v srdci a posiela krv do celého tela. Atriálna fibrilácia (AFib)nastáva, keď sa elektrický signál v predsieni stane chaotickým, a vedie k poruchám v intervale pulzu. AFib môžespôsobiť stagnáciu krvi v srdci, čím sa môžu ľahko vytvárať zrazeniny krvi, ktoré sú príčinou mozgovej mŕtvice a
srdcového infarktu.
Tlakovanie

Paličkový indikátor tlaku
Indikátor sleduje priebeh tlaku počas merania.

Prebieha nafukovanie
Vykonáva sa meranie
Nafukovanie dokončené

Slovenčina

12

Indikátor klasifikácie SZOKaždých šesť segmentov paličkového indikátora zodpovedá
klasifikácii SZO krvného tlaku, ktorá je opísaná na strane 14.
PríkladMierna hypertenzia Slabá hypertenzia

Indikátor klasifikácie SZOŤažká hypertenzia
Mierna hypertenzia
Slabá hypertenzia
Vysoký normálny
Normálny
Optimálny

Vysoký normálny: Indikátor zobrazuje segmentna základe aktuálnych údajov
zodpovedajúcich klasifikácii SZO.

Čo je krvný tlak?

O krvnom tlakuKrvný tlak je sila, ktorou krv pôsobí na steny tepien. Systolický tlak nastane, keď sa srdce stiahne. Diastolický tlak
nastane, keď srdce roztiahne. Krvný tlak sa meria v milimetroch ortuťového stĺpca (mmHg). Prirodzený krvný tlakčloveka predstavuje základný tlak nameraný hneď ráno, keď je človek stále v pokoji a pred jedlom.

Čo je hypertenzia a ako sa upravuje?

Hypertenzia, abnormálne vysoký arteriálny krvný tlak, ak sa nelieči, môže spôsobiť mnoho zdravotných problémov,vrátane mŕtvice a infarktu. Hypertenziu je možné upravovať zmenou životného štýlu, vyhýbaním sa stresu aužívaním liekov pod dohľadom lekára.
Predchádzanie hypertenzii alebo jej úprava:
Nefajčite,Pravidelne cvičte,Znížte príjem soli a tukov,
Podstupujte pravidelné lekárske prehliadky,Udržiavajte vhodnú hmotnosť.
Slovenčina

13

Prečo merať krvný tlak doma?

Krvný tlak meraný na klinike alebo v ordinácii môže vyvolať obavy a spôsobiť vyššie namerané hodnoty o 25 až
30 mmHg ako pri meraní doma. Meranie doma znižuje účinky vonkajších vplyvov na namerané hodnoty krvného
tlaku, dopĺňa hodnoty namerané lekárom a poskytuje presnejšiu a kompletnejšiu históriu krvného tlaku.

Štandardy na posúdenie vysokého krvného tlaku bez ohľadu
na vek boli stanovené Svetovou zdravotníckou organizáciou
(SZO), ako je znázornené v grafe vpravo.

Odchýlky krvného tlakuKrvný tlak človeka značne kolíše na dennom a sezónnom základe.
Môže sa líšiť o 30 až 50 mmHg v závislosti od rôznych podmienok v
priebehu dňa. V prípade ľudí s hypertenziou sú rozdiely eštevýraznejšie. Krvný tlak sa za normálnych okolností zvyšuje pri prácialebo pri hre a klesá na najnižšiu úroveň počas spánku. Takže sa
príliš neznepokojujte výsledkami jedného merania.
Meranie vykonávajte každý deň v rovnaký čas pomocou postupu
uvedeného v tomto návode, aby ste zistili, aký máte normálny krvný
tlak. Pravidelne merané hodnoty poskytujú úplnejšiu históriu
krvného tlaku. Pri zaznamenávaní svojho krvného tlaku
zaznamenajte aj dátum a čas. Pri interpretácii údajov o krvnom tlaku
sa poraďte s lekárom.

Slovenčina

14

Hypertenzia 3. stupňa (ťažká)Hypertenzia 2. stupňa (mierna)Hypertenzia 1. stupňa (slabá)
Vysoký – normálny
Normálny
Optimálny

Systolický krvný tlak
Typické odchýlky v rámci dňa(Merané každých päť minút)
Systolický

Krvný tlak

Klasifikácia SZO krvného tlaku

Diastolický krvný tlak

Referenčný materiál; Journal ofHypertension 1999, ročník 17 č. 2

Diastolický
Popoludní

SpánokPredpoludním Čas

Riešenie problémov
Problém

Možný dôvod
Sú vybité batérie.

Odporúčaná akcia
Vymeňte všetky batérie za nové.
Znova vložte batérie tak, aby ich záporný akladný koniec bol v súlade s označením v
priestore na batérie.

Na displeji sa nič
nezobrazuje, aj keď je
Konce batérií nie sú v správnej polohe.
zariadenie zapnuté.
Manžeta sa
nenafukuje.

Napätie batérií je príliš nízke.
Bliká symbol(TAKMER VYBITÁBATÉRIA). Ak sú batérie úplne vybité,
symbol sa nezobrazí.
Manžeta nie je nasadená správne.
Počas merania ste pohli zápästím alebo
telom.

Zariadenie nemeria.
Namerané hodnoty sú
Poloha manžety nie je správna.
príliš vysoké alebopríliš nízke.

Iné
Poznámka:

Hodnota je odlišná od hodnoty
nameranej na klinike alebo v ordinácii.

Vymeňte všetky batérie za nové.
Nasaďte manžetu správne.
Počas merania buďte úplne nehybní anerozprávajte.
Seďte pohodlne a nehybne.
Položte ruku na stôl dlaňou nahor a manžetou na
rovnakej úrovni ako vaše srdce.
Ak máte veľmi slabý alebo nepravidelný pulz
srdca, zariadenie môže mať ťažkosti s určením
krvného tlaku.
Pozri časť Prečo merať krvný tlak doma?.
Vyberte batérie. Vložte ich správne späť a skúste
zopakovať meranie.

Ak vyššie popísané akcie problém nevyriešia, obráťte sa na predajcu. Nepokúšajte sa otvárať alebo
opravovať tento výrobok, pretože v dôsledku akéhokoľvek takéhoto pokusu prestane platiť záruka.
Slovenčina

15Údržba
Neotvárajte zariadenie. Využíva citlivé elektrické súčasti a zložitú vzduchovú jednotku, ktorá by sa mohla poškodiť.
Ak problém nemôžete vyriešiť pomocou pokynov na riešenie problémov, obráťte sa na autorizovaného predajcu vo
svojom okolí alebo na naše oddelenie zákazníckeho servisu. Zákaznícky servis spoločnosti A&D poskytne
technické informácie, náhradné diely a jednotky autorizovaným predajcom.
Zariadenie bolo navrhnuté a vyrobené tak, aby malo dlhú životnosť. Vo všeobecnosti sa však odporúča, aby sa
zariadenie kontrolovalo každé 2 roky, aby sa zabezpečilo riadne fungovanie a presnosť. Informácie o údržbe
získate od autorizovaného predajcu vo vašom okolí alebo spoločnosti A&D.

Technické údaje
Typ
Metóda merania
Rozsah merania

Presnosť merania
Napájanie
Počet meraníObvod zápästia
Klasifikácia
Aplikovaný diel
Doba použiteľnosti

UB-525
Oscilometrické meranieTlak:
0 – 299 mmHg
Systolický tlak:
60 – 279 mmHg
Diastolický tlak:
40 – 200 mmHg
Pulz:
40 – 180 úderov/minúta
Tlak:±3 mmHgPulz:±5 %2 x 1,5 V alkalické batérie (LR03 alebo AAA)
Približne 250 meraní pri použití alkalických batérií AAA, hodnote tlaku 170 mmHg
pri teplote miestnosti 23 °C
13,5 – 21,5 cm
Vnútorne napájané zariadenie ME (Režim nepretržitej prevádzky)Manžeta Typ BF
Zariadenie: 5 rokov (ak sa používa šesťkrát denne)
Slovenčina 16

Klinické testy
EMC
Pamäť
Prevádzkové podmienky
Prepravné/skladovacie
podmienky
Rozmery
Hmotnosť
KrytiePodľa ISO81060-2：2013
V klinickej overovacej štúdii sa K5 použilo na 85 subjektoch na určenie diastolického
krvného tlaku.
IEC 60601-1-2: 2014
Posledných 60 meraní+10 až +40 °C/15 až 85 % RV/800 až 1 060 hPa-20 až +60 °C/10 až 95 % RV/700 až 1 060 hPa
Približne 56 [D] x 88 [V] x 21,5 [H] mm
Približne 90 g, bez batérií
IP20

Poznámka: Technické parametre sa môžu zmeniť bez predchádzajúceho upozornenia.
Klasifikácia IP je stupeň ochrany poskytovaný krytmi podľa normy IEC 60529. Toto zariadenie je chránené
proti pevným cudzím telesám s priemerom 12 mm a väčším, ako napríklad prst. Toto zariadenie nie jechránené voči vode.

Slovenčina

17

POZNÁMKA

Slovenčina

18

Vsebina

Spoštovane stranke ............................................................................................................................................. 2
Začetne opombe .................................................................................................................................................. 2
Previdnostni ukrepi .............................................................................................................................................. 2
Identifikacija delov ............................................................................................................................................... 4
Simboli ................................................................................................................................................................. 5
Uporaba zaslona.................................................................................................................................................. 7
1. Vstavljanje/odstranjevanje baterij ............................................................................................................... 7
2. Nameščanje manšete ................................................................................................................................ 8
3. Kako izvajati natančna merjenja................................................................................................................. 8
4. Merjenje ..................................................................................................................................................... 9
5. Po merjenju ................................................................................................................................................ 9
Merjenja ............................................................................................................................................................. 10
Priklic podatkov iz pomnilnika ............................................................................................................................ 11
Brisanje vseh podatkov, shranjenih v pomnilniku .............................................................................................. 11
Kaj je indikator IHB/AFib? .................................................................................................................................. 12
Kaj je atrijska fibrilacija? .................................................................................................................................... 12
Vrstični indikator tlaka ........................................................................................................................................ 12
Indikator razvrstitve SZO ................................................................................................................................... 13
O krvnem tlaku................................................................................................................................................... 13
Kaj je krvni tlak? ........................................................................................................................................... 13
Kaj je povišan krvni tlak in kako ga nadzorujemo?....................................................................................... 13
Zakaj si krvni tlak merimo doma? ................................................................................................................. 14
Razvrstitev krvnega tlaka SZO ..................................................................................................................... 14
Spreminjanje krvnega tlaka .......................................................................................................................... 14
Odpravljanje težav ............................................................................................................................................. 15Vzdrževanje ....................................................................................................................................................... 16
Tehnični podatki................................................................................................................................................. 16

Slovenščina

1

Spoštovane strankeČestitamo vam za nakup vrhunske naprave za merjenje krvnega tlaka A&D. Ta naprava, ki je zasnovana za
enostavno uporabo in natančnost, bo olajšala vaš dnevni postopek merjenja krvnega tlaka.
Priporočamo, da pred prvo uporabo te naprave temeljito preberete ta priročnik.

Začetne opombe
Ta naprava je skladna z Evropsko direktivo 93/42 EGS o medicinskih izdelkih. To jasno označuje oznaka o. (0123: referenčna številka vključenega priglašenega organa)skladnosti
Naprava je zasnovana za uporabo na odraslih osebah, ne na novorojenčkih ali dojenčkih.
Okolje za uporabo. Naprava je zasnovana za samostojno uporabo v domačem zdravstvenem okolju.
Ta naprava je zasnovana za merjenje krvnega tlaka in srčnega utripa ljudi za diagnostične namene.

Previdnostni ukrepi

Pri izdelavi te naprave so bili uporabljeni natančni sestavni deli. Izogibati se je treba visoki in nizki temperaturi, vlagi,
neposredni sončni svetlobi, močnim udarcem ali veliki količini prahu.
Napravo očistite s suho, mehko krpo ali krpo, navlaženo z vodo in nevtralnim čistilnim sredstvom. Za čiščenje
naprave nikoli ne uporabljajte alkohola, benzena, razredčila ali drugih močnih kemikalij.
Izogibajte se tesnemu zlaganju manšete za daljše obdobje, saj lahko takšna dejanja skrajšajo življenjsko dobo
sestavnih delov.
Naprava ni vodoodporna. Preprečite, da bi dež, pot in voda umazali napravo.
Merjenja so lahko popačena, če se naprava uporablja v bližini televizorjev, mikrovalovnih pečic, mobilnih telefonov,rentgenskih ali drugih naprav z močnimi električnimi polji.
Rabljena oprema, deli in baterije ne smejo biti obravnavani kot običajni gospodinjski odpadki in jih je treba odstraniti
v skladu z veljavnimi lokalnimi uredbami.
Ko ponovno uporabljate napravo, se prepričajte, da je čista.
Ne spreminjajte naprave. To lahko povzroči nesreče ali poškodbe naprave.
Za merjenje krvnega tlaka mora manšeta zapestje stisniti dovolj čvrsto, da se začasno zaustavi pretok krvi skozi
arterijo. To lahko povzroči bolečino, otrplost ali začasno rdečino na zapestju. To stanje se bo še posebej pojavilo v
primeru zaporednih ponovitev merjenja. Možne bolečine, otrplost ali rdečine bodo sčasoma izginile.
Brezžična komunikacijska naprava, npr. domače omrežne naprave, mobilni telefoni, brezžični telefoni in njihove
Slovenščina 2

postaje, baterijski sprejemniki in oddajniki, lahko vplivajo na to napravo za merjenje krvnega tlaka. Zato je treba
zagotoviti najmanj 30 cm razdalje od teh naprav.
Prepogosto merjenje krvnega tlaka lahko povzroči poškodbe zaradi motenj krvnega pretoka. Preverite, dadelovanje
naprave ne povzroči daljše motnje krvnega obtoka, če napravo uporabljate večkrat zaporedoma.
Klinični testi niso bili izvedeni na novorojenčkih in nosečnicah. Naprave ne uporabljajte na novorojenčkih ali
nosečnicah.
Če je bila na vas izvedena amputacija dojk, se pred uporabo naprave posvetujte z zdravnikom.
Otrokom ne dovolite samostojne uporabe naprave in naprave ne uporabljajte v prostoru, kjer bi bila na dosegu
dojenčkov. To lahko povzroči nesreče ali poškodbe.
Naprava vsebuje majhne dele, ki predstavljajo nevarnost zadušitve, če jih pogoltnejo dojenčki.
Istočasno se ne dotikajte baterij in bolnika. To lahko povzroči električni udar.
V primeru okvare posameznih sestavnih delov se lahko ohišje manšete ogreje in povzroči morebitno škodo.
Uporaba dodatkov, ki niso opisani v tem priročniku, lahko ogrozi varnost.
Če pride do kratkega stika baterije, se lahko ta segreje in povzroči opekline.
Pred uporabo omogočite, da se naprava prilagodi okolju (približno eno uro).
Manšete ne napihujte, dokler je ne ovijete okrog zapestja.

Kontraindikacije

V nadaljevanju so navedeni previdnosti ukrepi za primerno uporabo naprave.
Naprave ne uporabljate na zapestju, kjer že uporabljate drugo medicinsko opremo. Oprema morda ne bo delovala
primerno.
Ljudje, ki imajo resne motnje obtoka v roki, se morajo pred uporabo naprave posvetovati z zdravnikom, da seizognejo medicinskim težavam.
Ne izvajajte samodiagnoze rezultatov merjenja in ne začenjajte samostojnega zdravljenja. Glede vrednotenja
rezultatov in zdravljenja se vedno posvetujte z zdravnikom.
Naprave ne uporabljajte na roki, na kateri je prisotna nezaceljena rana.
Naprave ne uporabljajte na roki, na kateri se izvaja intravenozna infuzija ali transfuzija krvi. To lahko povzroči
poškodbe ali nesreče.
Naprave ne uporabljajte v prisotnosti vnetljivih plinov, npr. anestetičnih plinov. To lahko povzroči eksplozijo.
Naprave ne uporabljajte v okoljih z visoko koncentracijo kisika, npr. v visokotlačni kisikovi komori ali kisikovem
šotoru. To lahko povzroči požar ali eksplozijo.
Slovenščina

3

Identifikacija delov
Zaslongumb (povprečje, pomnilnik in zamenjava prednastavljenih parametrov)

Gumb START
Pokrov predela zabaterije

Manšeta

POMNILNIK
Povprečje

ZaslonAlkalne baterije (LR03 ali AAA)

Sistolični tlak
Diastolični tlak

Indikator razvrstitve SZO
in
Vrstični indikator tlaka

Simbol IHB/AFibOznaka za srce

Srčni utrip

Indikator za baterije
Slovenščina

4

Simboli

Simboli, ki so natisnjeni na ohišju naprave
Simboli

Funkcija/pomen

V pripravljenosti in vklop naprave
SYS

Sistolični krvni tlak v mmHg

DIA

Diastolični krvni tlak v mmHg

PUL

Utripov na minuto
Vodič za namestitev baterij
Enosmerni tok

SN

Serijska številka
Datum proizvodnje
Tip BF: naprava in manšeta sta zasnovani za zagotavljanje posebne zaščite pred električnimi udari.
Oznaka medicinske naprave glede Direktive ES

IP

Mednarodni simbol za zaščito
Oznaka OEEO
Proizvajalec
Predstavnik EU
Glejte priročnik za uporabo/priročnik
Ohranite suho

Slovenščina

5

Simboli, ki se pojavijo na zaslonu
Simboli

Funkcija/pomen/priporočen ukrep
Pojavi se, ko poteka merjenje.Utripa, ko je zaznan utrip.
Ostanite čim bolj pri miru.
Simbol IHB/AFib se pojavi, ko je zaznan nepravilen srčni utrip.
Lahko zasveti, ko je zaznana rahla vibracija, npr. drhtenje ali tresenje.

Simboli, ki se pojavijo na zaslonu (trajno)
Simboli

Funkcija/pomen

Priporočen ukrep

Predhodna merjenja, shranjena v POMNILNIKU.
Povprečni podatki
POLNA BATERIJA
Indikator stanja baterij med merjenjem
NIZKO STANJE NAPOLNJENOSTI BATERIJE
Ko utripa, so baterije skoraj prazne
Nestabilen krvni tlak zaradi premikanja med merjenjemali

Ko indikator utripa, vse baterije
zamenjajte z novimi.
Izvedite še eno merjenje.
Med merjenjem bodite pri miru.

Sistolična in diastolična vrednost se nahajata znotraj 10 mmHg
ena od druge.
Vrednost tlaka se med napihovanjem ni povečala.
Manšeta ni pravilno nameščena.
NAPAKA PRIKAZA PUL.
Utrip ni pravilno zaznan.

Pravilno pritrdite manšeto in izvedite
še eno merjenje.

Notranja napaka naprave za merjenje krvnega tlaka

Odstranite baterije in pritisnite gumb
gumb START , potem pa ponovno
vstavite baterije. Če se napaka
ponovno pojavi, se obrnite na
prodajalca.

Slovenščina

6

Uporaba zaslona

1. Vstavljanje/odstranjevanje baterij
1. Odstranite pokrov predela za baterije.

Pokrov predela
za baterije

2. Odstranite uporabljene baterije in v skladu s prikazom vstavite
nove baterije v predel za baterije, pri čemer pazite na pravilne
polarnosti (+ in-) Uporabljajte samo baterije LR03 ali AAA.
3. Namestite pokrov predela za baterije.

Uporabljenebaterije

POZOR

Baterije vstavite v predel za baterije, kot je prikazano.
Nove baterije
Če baterije vstavite nepravilno, naprava ne bo delovala.
Ko na zaslonu utripa(oznaka NIZKO STANJE
NAPOLNJENOSTI BATERIJE), vse baterije zamenjajte z
novimi. Ne pomešajte starih in novih baterij.To lahko skrajšaživljenjsko dobo baterij ali povzroči nepravilno delovanje
naprave.
(oznaka NIZKO STANJE NAPOLNJENOSTIBATERIJE) se ne pojavi, ko so baterije povsem prazne.
Življenjska doba baterij se lahko razlikuje glede na temperaturo okolice in je lahko v primeru nizkih temperatur
krajša. Na splošno bosta dve novi bateriji LR03/AAA zdržali približno štiri mesece, če se naprava uporablja za dve
merjenji na dan.
Uporabljajte samo navedene baterije. Baterije, ki so bile dobavljene z napravo, so namenjene za preverjanjedelovanja naprave in imajo morda omejeno življenjsko dobo.
Če naprave dlje časa ne boste uporabljali, odstranite baterije. Baterije lahko puščajo in povzročijo napačno
delovanje naprave.

Slovenščina

7

2. Nameščanje manšete
1. Manšeto namestite okrog zapestja približno 1 cm nad roko, kot je prikazano na
sliki na desni.

Približno 1 cm

2. Manšeto pritrdite z velcro trakom.
Opomba: Za točno merjenje, manšeto namestite trdno in merjenje opravite na
golem zapestju.

3. Kako izvajati natančna merjenja

Za najbolj natančna merjenja krvnega tlaka:
Med merjenjem mirujte in ne govorite.
Usedite se v udoben položaj. Komolec položite na mizo, pri čemer naj bo vaša dlan obrnjena navzgor, manšeta pa
naj bo na višini vašega srca.
Pred merjenjem se umirite za pet do deset minut. Če ste vznemirjeni ali depresivni zaradi emocionalnega stresa,
potem se bo ta stres pri merjenju pokazal kot višji (ali nižji) krvni tlak (v primerjavi z običajnim krvnim tlakom), srčni
utrip pa bo običajno hitrejši od normalnega.
Krvni tlak si poskušajte izmeriti vsak dan ob približno isti uri.
Krvni tlak posameznika se stalno spreminja, odvisno od tega kaj počnete in kaj ste jedli. Tisto, kar popijete, ima
lahko zelo močan in hiter učinek na vaš krvni tlak.
Merjenja ne izvajajte takoj po fizični dejavnosti ali kopeli. Pred izvedbo merjenja počivajte dvajset ali trideset minut.
Ne prekrižajte nog. Noge ohranite plosko na tleh in izravnajte hrbet.
Meritve te naprave temeljijo na srčnem utripu. Če imate zelo slaboten ali nepravilen srčni utrip, potem bo naprava
morda imela težave pri določanju vašega krvnega tlaka.
Če bo naprava zaznala nenormalno stanje, bo zaustavila merjenje in prikazala simbol napake. Za opise simbolov
glejte stran 6.
Naprava za merjenje krvnega tlaka je namenjena odraslim. Pred uporabo naprave na otroku se posvetujte s svojim
zdravnikom. Otrok te naprave ne sme uporabljati brez nadzora.
Na delovanje samodejne naprave za merjenje krvnega tlaka lahko vplivajo prekomerna temperatura, vlaga ali
nadmorska višina.
Slovenščina 8

4. Merjenje

Med merjenjem je normalno, če se zdi manšeta zelo tesna.

5. Po merjenju

Če med prikazom meritev pritisnete gumb START , da izklopite napravo, se bodo nove izmerjene vrednosti
shranile v pomnilnik.
Če med prikazom meritev pritisnete gumb , da izklopite napravo, se izmerjene vrednosti ne bodo shranile v
pomnilnik.
Odstranite manšeto in posnemite vaše podatke.
Opomba: Naprava ima funkcijo samodejnega izklopa napajanja, ki shrani trenutne podatke v pomnilnik in napravo
izklopi približno eno minuto po meritvi. Pri merjenjih na isti osebi med merjenji počakajte vsaj tri minute.

Slovenščina

9

Merjenja
Opomba: UB-525 bo ob uporabi zagotovil napihovanje, ki je primerno za uporabnika.
1. Manšeto ovijte okrog zapestja. Udobno sedite in središče manšete postavite navišino vašega srca ter se sprostite.
2. Pritisnite gumb START . Vsi segmenti zaslona so prikazani.
3. Ničla (0) utripa za kratek čas. Ko se merjenje začne, se zaslon spremeni.
Manšeta se začne napihovati. Normalno je, če se zdi manšeta zelo tesna. Ko senapihovanje začne, in utripa(oznaka srca), se merjenje samodejno zažene.
Opomba: Če želite kadar koli zaustaviti napihovanje, ponovno pritisnite
gumb START .
4. Ko je merjenje dokončano, naprava prikaže odčitke (odčitka sistoličnega indiastoličnega tlaka, srčni utrip, razvrstitev SZO in simbol I.H.B.). Manšeta izpusti
preostali zrak in se samodejno povsem izprazni.
Opomba: Če novih odčitkov ne želite shraniti v pomnilnik, med prikazovanjem
odčitkov pritisnite gumb .
5. Ponovno pritisnite gumb START , da izklopite napravo. Odstranite
manšeto.
Opomba: Naprava ima funkcijo samodejnega izklopa napajanja.
Pri merjenjih na isti osebi med merjenji počakajte vsaj tri minute.

Slovenščina

10

Vsi segmenti

Brez prikaza
Napihovanje se začne
Napihovanje inmerjenje potekata
Sistolični tlak
Diastolični tlak
Razvrstitev SZOSimbol I.H.B.
Srčni utrip

Priklic podatkov iz pomnilnika
Opomba:Ta naprava v pomnilnik shrani zadnjih 60 merjenj.
1. Ponovno pritisnite gumb , da izklopite napravo.
Prikaže se povprečje vseh merjenj in število podatkov. Če ni podatkov, seprikaže »0«. Pritisnite gumbali START , da izklopite napravo.
2. Uporabite naslednje gumbe, da prikažete podatke (število podatkov in
podatke merjenja).
Naprava prikaže povprečje vseh merjenj in število podatkov.
S ponovnim pritiskom gumba , naprava prikaže naslednje:
Povprečne podatke vseh vmesnih merjenj.
Podatki (število podatkov in podatki merjenja).
Naprava prikaže podatke razvrščene po aktualnosti.
Na primer:
št. 30 in podatki
št. 29 in podatki
št. 01in podatki.
3. Če pritisnete gumb , ko se prikažejo najstarejši podatki, napravanadaljuje s korakom 1 in prikaže povprečje vseh merjenj in število
podatkov.
4. Ponovno pritisnite gumb START , da izklopite napravo.
Po eni minuti nedelovanja se bo naprava samodejno izklopila.

Način pomnilnika

gumb
Povprečje vseh
podatkovŠtevilo vseh podatkov, briše podatke, shranjene v pomnilniku, medtem
Naprava prikaže oznako
utripa in se nato samodejno izklopi.
ko oznaka
Slovenščina

11

gumb

Primer: št. 30
Najnovejši podatki
gumb
Zadnji podatki(najstarejši)

Brisanje vseh podatkov, shranjenih v pomnilniku
1. Pritisnite in zadržite gumb , da se naprava samodejno izklopi.

Izklopgumb

IzklopgumbKaj je indikator IHB/AFib?
Ko naprava med merjenji zazna nepravilen ritem, se na zaslonu z vrednostmi merjenja prikaže indikator IHB/AFib.
Opomba: Če se med merjenjem pogosto pojavlja ta
svojega zdravnika.

indikator IHB/AFib, priporočamo, da se obrnete na

Kaj je atrijska fibrilacija?

Srce se krči zaradi električnih signalov, ki se pojavljajo v srcu, in pošilja kri po telesu. Atrijska fibrilacija (AFib) se
zgodi, ko pride do napake električnega signala v atriju in takšen signal potem privede do motenj v intervalu utripa.
Atrijska fibrilacija povzroči stagnacijo krvi v srcu, kar lahko hitro privede do krvnih strdkov, ki lahko potem povzročijo
kap ali srčni infarkt.
Ustvarjanje tlaka

Vrstični indikator tlaka

Indikator spremlja napredek tlaka med merjenjem.
Napihovanje poteka
Merjenje poteka
Napihovanje dokončano

Slovenščina

12

Indikator razvrstitve SZO
Vsak šesti segment vrstičnega indikatorja se sklada z
razvrstitvijo krvnega tlaka SZO, opisano na strani 14.
PrimerZmerno povišan
krvni tlak

Rahlo povišan krvni
tlak

Indikator razvrstitve SZO
Resno povišan krvni tlak
Zmerno povišan krvni tlak
Rahlo povišan krvni tlak
Visok-normalen
Normalen
Optimalen

Visok-normalen: Indikator prikazuje segment na podlagi trenutnih
podatkov, ki se sklada z razvrstitvijo SZO. na
podlagi trenutnih podatkov, ki se sklada z
razvrstitvijo SZO.

Kaj je krvni tlak?

O krvnem tlaku

Krvni tlak je sila, ki jo kri izvaja na stene arterij. Sistolični tlak nastane, ko se srce krči. Diastolični tlak nastane, ko sesrce širi. Krvni tlak se meri v milimetrih srebra (mmHg). Naravni krvni tlak predstavlja osnovni krvni tlak, ki ga
izmerimo zgodaj zjutraj, ko smo še spočiti in še nismo jedli.

Kaj je povišan krvni tlak in kako ga nadzorujemo?

Povišan krvni tlak, nenormalno visok arterijski krvni tlak lahko, če ga ne zdravimo, povzroči številne zdravstvenetežave, vključno s kapjo in srčnim infarktom. Povišan krvni tlak lahko nadzorujemo s spremembo življenjskega
sloga, izogibanjem stresu in zdravili pod nadzorom zdravnika.
Za preprečevanje visokega krvnega tlaka ali njegovo nadzorovanje:
Ne kaditeRedno telovaditeZnižajte vnos soli in maščob
Hodite na redne zdravniške pregledeVzdržujte primerno telesno težoSlovenščina 13

Zakaj si krvni tlak merimo doma?

Merjenje krvnega tlaka v kliniki ali pri zdravniku lahko povzroči strah, zaradi česar je lahko odčitek krvnega tlaka višji,
in sicer tudi od 25 mmHg do 30 mmHg višji od tistega, ki ga izmerite doma. Merjenje krvnega tlaka doma zmanjša
zunanje učinke na odčitke krvnega tlaka, dopolnjuje odčitke zdravnika in zagotavlja bolj natančno, celovito
zgodovino krvnega tlaka.
Referenčno gradivo; Journal of

Razvrstitev krvnega tlaka SZO

Spreminjanje krvnega tlaka

Krvni tlak posameznika se močno razlikuje glede na dan in letni čas.
Zaradi različnih pogojev tekom dneva se lahko razlikuje tudi za
30–50 mmHg. Pri posameznikih s povišanim krvnim tlakom so tespremembe še bolj poudarjene. Običajno se krvni tlak dvigne med
delom ali igro, med spanjem pa pade na najnižje ravni. Zato ne
bodite preveč zaskrbljeni zaradi rezultata enega merjenja.
Ob upoštevanju postopka, ki je opisan v tem priročniku, merjenja
izvajajte vsak dan ob istem času, da pridobite podatke o svojem
normalnem krvnem tlaku. Z rednimi odčitavanji ustvarite boljobsežno zgodovino krvnega tlaka. Pri merjenju svojega krvnegatlaka zabeležite tudi datum in čas. Za razlago podatkov o vašem
krvnem tlaku se posvetujte s svojim zdravnikom.

Slovenščina

14

Povišan krvni tlak 3. stopnje (resen)
Povišan krvni tlak 2. stopnje (zmeren)
Povišan krvni tlak 1. stopnje (rahel)
Visok-normalen
Normalen
Optimalen

Sistolični krvni tlak
Običajne spremembe v enem dnevu(merjeno vsakih pet minut)
Sistolični
Krvni tlak

Standarde za ocenjevanje visokega krvnega tlaka, ne glede na
starost, je zastavila Svetovna zdravstvena organizacija (SZO),
kot je prikazano na diagramu na desni.

Diastolični krvni tlak

Hypertension 1999, izdaja 17, št. 2

Diastolični
PopoldanSpanjeDopoldanČasOdpravljanje težav
TežavaNa zaslonu se ne
pojavi nič, tudi ko jenaprava vklopljena.

Možen vzrok
Baterije so prazne.
Poli baterij niso v pravilnem položaju.

Priporočen ukrep
Vse baterije zamenjajte z novimi.
Ponovno vstavite baterije tako, da se negativni inpozitivni poli skladajo s tistimi, ki so označeni na
predelu za baterije.

Napetost baterije je prenizka.
(oznaka NIZKO STANJE
Manšeta se ne
Vse baterije zamenjajte z novimi.
NAPOLNJENOSTI BATERIJE) utripa.
napihne.
Če so baterije povsem prazne, se oznaka
ne prikaže.
Manšeta ni primerno nameščena.
Pravilno namestite manšeto.
Med merjenjem ste premaknili zapestje
Med merjenjem bodite čim bolj pri miru in ne
ali trup.
govorite.
Naprava ne izvede
Sedite udobno in pri miru.
merjenja.
Roko položite na mizo, pri čemer naj bo vašaPoložaj manšete ni pravilen.
Odčitki so previsoki ali
dlan obrnjena navzgor, manšeta pa naj bo na
prenizki.
višini vašega srca.
Če imate zelo slaboten ali nepravilen srčni utrip,
bo naprava morda imela težave pri določanju
vašega krvnega tlaka.
Vrednost je drugačna od tiste, ki je bila
Glejte razdelek »Zakaj si krvni tlak merimo
izmerjena v kliniki ali pri zdravniku.
doma?«.
Drugo
Odstranite baterije. Vstavite jih pravilno in
ponovno izvedite merjenje.
Opomba: Če zgoraj opisani ukrepi ne razrešijo težave, se obrnite na prodajalca. Ne poskušajte odpirati ali sami
popraviti izdelka, saj bo v tem primeru prišlo do razveljavitve garancije.
Slovenščina 15

Vzdrževanje
Ne odpirajte naprave. Naprava uporablja občutljive električne sestavne dele in zapleteno zračno enoto, ki se lahko
poškodujejo. Če težave ne morete odpraviti s pomočjo navodil za odpravljanje težav, se obrnite na pooblaščenega
prodajalca v vašem območju ali na oddelek za pomoč strankam. Oddelek za pomoč strankam podjetja A&D bo
pooblaščenim prodajalcem zagotovil tehnične informacije, rezervne dele in enote.
Naprava je bila zasnovana in proizvedena za dolgo življenjsko dobo. Vendar pa na splošno priporočamo, da se
vsaki 2 leti na napravi izvede pregled, s čimer se zagotovita primerno delovanje in natančnost. Prosimo, da seglede vzdrževanja obrnete na pooblaščenega prodajalca v vašem območju ali na podjetje A&D.

Tehnični podatki
Tip
Metoda merjenja
Razpon merjenja

Natančnost merjenja
NapajanjeŠtevilo merjenj
Obod zapestja
Razvrstitev
Uporabljen delŽivljenjska doba

UB-525
Oscilometrično merjenje
Tlak:
0 do 299 mmHg
Sistolični tlak: 60 do 279 mmHg
Diastolični tlak: 40 do 200 mmHg
Srčni utrip:
40 do 180 utripov/minuto
Tlak:±3 mmHg
Utrip:±5%
2 x 1,5 V alkalne baterije (LR03 ali AAA)
Pribl. 250 merjenj, pri uporabi alkalnih baterij AAA, z vrednostjo tlaka 170 mmHg pri
sobni temperaturi 23 °C.
13,5 – 21,5 cm
Oprema ME z notranjim napajanjem (trajni način delovanja)
Manšeta tipa BF
Naprava: 5 let (če se uporablja šestkrat na dan)Slovenščina

16

Klinični test
EMD
Pomnilnik
Pogoji delovanja
Transport/pogoji shranjevanja
DimenzijeTeža
Zaščita pred vdorom

V skladu s standardom ISO81060-2：2013
V klinični validacijski študiji so za določanje diastoličnega krvnega tlaka vrednost K5
uporabili pri 85 osebah.
IEC 60601-1-2: 2014
Zadnjih 60 merjenj+10 do +40 °C / 15 do 85 % RH / 800 do 1.060 hPa-20 do +60 °C / 10 do 95 % RH / 700 do 1.060 hPa
Pribl. 56 [W] x 88 [H] x 21,5 [D] mm
Pribl. 90 g, brez baterij
IP20

Opomba: Tehnični podatki so lahko predmet izboljšav brez predhodnega obvestila.
Razvrstitev IP predstavlja stopnje zaščite, ki jih zagotavljajo ohišja v skladu s standardom IEC 60529.
Naprava je zaščitena pred trdimi tujki premera 12 mm in več, kot so npr. prsti. Ta naprava ni zaščitena predvodo.

Slovenščina

17

MEMO

Slovenščina

18

Tartalomjegyzék

Tisztelt felhasználó! ............................................................................................................................................. 2
Használat előtti megjegyzések ............................................................................................................................ 2Óvintézkedések ................................................................................................................................................... 2
Az eszköz részei .................................................................................................................................................. 4
Szimbólumok ....................................................................................................................................................... 5
A vérnyomásmérő használata ............................................................................................................................. 7
1. Az elemek behelyezése/cseréje ................................................................................................................. 7
2. A mandzsetta felhelyezése ........................................................................................................................ 8
3. Hogyan végezhetők pontos mérések? ....................................................................................................... 8
4. Mérés közben ............................................................................................................................................. 9
5. Mérés után ................................................................................................................................................. 9
Mérések ............................................................................................................................................................. 10
A memóriában tárolt adatok megtekintése ........................................................................................................ 11
A memóriában tárolt összes adat törlése .......................................................................................................... 11
Mit jelent a szabálytalan szívritmus/pitvarfibrilláció jelzése? ............................................................................. 12
Mi az a pitvarfibrilláció? ..................................................................................................................................... 12
Vérnyomásjelző ................................................................................................................................................. 12
WHO szerinti osztályozás jelzője ....................................................................................................................... 13
Tudnivalók a vérnyomásról ................................................................................................................................ 13
Mi az a vérnyomás? ..................................................................................................................................... 13
Mi az a hipertónia, és hogyan szabályozható? ............................................................................................ 13
Miért jó az otthoni vérnyomásmérés? .......................................................................................................... 14
A vérnyomásértékek osztályozása a WHO szerint ...................................................................................... 14
A vérnyomás változása ................................................................................................................................ 14
Hibaelhárítás...................................................................................................................................................... 15
Karbantartás ...................................................................................................................................................... 16Műszaki adatok .................................................................................................................................................. 16

Magyar1

Tisztelt felhasználó!
Köszönjük, hogy az A&D korszerű vérnyomásmérőjét választotta. A könnyű használhatóság és a pontosság jegyében
tervezett eszköz megkönnyíti a napi vérnyomásmérési feladatokat.
Javasoljuk, hogy az eszköz használatának megkezdése előtt figyelmesen olvassa végig ezt a használati útmutatót.

Használat előtti megjegyzések

Ez az eszköz megfelel az orvostechnikai eszközökkel kapcsolatos 93/42 EGK európai irányelvnek. Ezt amegfelelőségi jelölés egyértelműen jelzi. (0123: Az érintett bejelentett szervezethez tartozó hivatkozási szám)
Az eszköz felnőttek számára készült, újszülöttek és csecsemők esetében nem használható.
Felhasználási környezet: az eszköz önállóan, otthoni egészségügyi ellátásban használható.
Az eszköz diagnosztikai célú vérnyomás - és pulzusmérésre alkalmas.

Óvintézkedések

Az eszköz precíziós alkatrészek felhasználásával készült. Ne tegye ki extrém hőmérsékletnek és páratartalomnak,közvetlen napfénynek, ütődésnek, valamint pornak.
Az eszköz tisztítását száraz, puha vagy nedves, semleges tisztítószeres ronggyal végezze. Ne használjon alkoholt, benzolt,
hígítót vagy egyéb erős hatású vegyszert az eszköz tisztításához.
Ne tárolja hosszú ideig szorosan összehajtva a mandzsettát, mert az az élettartam rövidülését okozhatja.
Az eszköz nem vízálló. Ügyeljen rá, hogy eső, izzadság vagy víz ne áztassa át.
A mérési eredményeket befolyásolhatja, ha az eszközt televíziókészülék, mikrohullámú sütő, mobiltelefon,
röntgenberendezés vagy más, erős elektromos mezővel rendelkező eszköz közelében használja.
Az elhasználódott eszköz, alkatrészek és akkumulátorok nem minősülnek általános háztartási hulladéknak, ezért azártalmatlanításukat a vonatkozó helyi jogszabályoknak megfelelően kell végezni.
Az eszköz ismételt használata előtt ellenőrizze, hogy az eszköz tiszta-e.
Ne alakítsa át az eszközt. Az eszköz átalakítása balesetet okozhat, vagy az eszköz károsodásához vezethet.
A vérnyomás mérésekor a mandzsettának elegendő mértékben szorítania kell a csuklót ahhoz, hogy az artériábanidőlegesen elzárja a véráramlást. Ez fájdalmat, zsibbadást okozhat, vagy ideiglenesen piros nyomot hagyhat a csuklón. Az
említett tünetek különösen a többszörösen ismételt mérések után jelentkezhetnek. Az ebből eredő fájdalom, zsibbadás vagy
pirosodás idővel elmúlik.

Magyar2

A vezeték nélküli kommunikációs eszközök, például az otthoni hálózati eszközök, mobiltelefonok, vezeték nélküli telefonok
és dokkolók, illetve a kézi adóvevők megzavarhatják a vérnyomásmérő működését. Ezért azt az ilyen eszközöktől legalább
30 cm távolságra kell használni.
A vérnyomás túlzottan gyakori mérése a vér áramlásának megzavarása miatt ártalmas lehet. Ellenőrizze, hogy az eszközműködése ismételt használat esetén nem okoz-e hosszú távú keringési zavart.
Az eszköz klinikai tesztelésében újszülöttek és terhes nők nem vettek részt. Újszülöttek és terhes nők esetében ne használja.
Ha emlőeltávolító műtéten esett át, az eszköz használata előtt konzultáljon kezelőorvosával.
Ne hagyja, hogy az eszközt gyermekek felügyelet nélkül használják, és tárolja csecsemőktől elzárva. Az eszköz balesetet
vagy sérülést okozhat.
Az apró alkatrészek miatt csecsemőknél véletlen lenyelés esetén fulladás veszélye áll fenn.
Ne érjen egyszerre az elemekhez és a pácienshez, mert az áramütést eredményezhet.
Egy alkatrész meghibásodása esetén a mandzsetta körüli burkolat felforrósodhat, és ez az eszköz meghibásodását okozhatja.
A jelen használati útmutatóban nem szereplő kiegészítők használata biztonsági kockázattal járhat.
Az elemek rövidzárlat esetén felforrósodhatnak, és égési sérülést okozhatnak.
Az eszköz használata előtt várjon körülbelül egy órát, hogy az alkalmazkodhasson a környezeti viszonyokhoz.
Ne fújja fel a mandzsettát, ha az nincs felhelyezve a csuklóra.

Ellenjavallatok

Az alábbiakban az eszköz helyes használatára vonatkozó óvintézkedések olvashatók.
Ne használja a mandzsettát olyan csuklón, amelyhez más elektromos orvosi berendezés csatlakozik, mivel az
megzavarhatja a berendezés működését.
A súlyos kari keringési rendellenességben szenvedőknek az egészségügyi problémák elkerülése érdekében az eszköz
használata előtt konzultálniuk kell kezelőorvosukkal.
Ne állítson fel öndiagnózist a mérési eredményekből, és ne kezdjen meg önállóan semmilyen kezelést. Az eredmények
kiértékeléséért és a kezelési módszer megállapításáért forduljon kezelőorvosához.
Ne használja az eszközt olyan csuklón, amelyen be nem gyógyult seb található.
Ne használja az eszközt olyan karon, amelyen keresztül a páciens intravénás infúziót vagy vérátömlesztést kap. Ez
balesetet vagy egészségkárosodást okozhat.
Ne használja az eszközt gyúlékony gázok (például altatógáz) közelében, mivel fennáll a robbanás veszélye.
Ne használja az eszközt magas oxigénkoncentrációjú környezetben, például magasnyomású oxigénkamrában vagy
oxigénsátorban, mert tüzet vagy robbanást okozhat.

Magyar3

Az eszköz részei
Kijelző

gomb (átlag, memória és az előre beállított paraméterek módosítása)

START gomb

Elemtartó rekesz fedele

MandzsettaMEMÓRIAÁtlag

Alkáli elemek (LR03 vagy AAA)

Kijelző
Szisztolés vérnyomás
Diasztolés nyomás

WHO szerinti osztályozás jelzője
és
Vérnyomásjelző
Szabálytalan szívritmus/pitvarfibrilláció szimbóluma

Szívdobogás szimbóluma

Pulzusszám

Elemkapacitás-jelző

Magyar4

Szimbólumok

Az eszközön szereplő szimbólumok
Szimbólumok

Funkció/jelentés

Készenlét és az eszköz bekapcsolása
SYS

Szisztolés vérnyomás (mmHg)

DIA

Diasztolés vérnyomás (mmHg)

PUL

Pulzusszám/perc
Az elemek behelyezési iránya
Egyenáram

SN

Sorozatszám
Gyártás dátuma
BF típusú: Az eszköz és a mandzsetta áramütés elleni védelemmel van ellátva
EK-irányelveknek megfelelő orvosi eszköz címkeIPIP-védettség szimbólumaWEEE címke
Gyártó
EU-képviselet
Lásd a használati útmutatót/kézikönyvet
Száraz helyen tartandó

Magyar5

A kijelzőn megjelenő szimbólumok
Szimbólumok

Funkció/jelentés/javasolt teendő
Mérés közben jelenik meg. Villog, ha pulzus észlelhető.
Maradjon mozdulatlan, amennyire csak lehet.
Szabálytalan szívritmus/pitvarfibrilláció szimbóluma. Akkor jelenik meg, ha az eszköz szabálytalan szívritmust észlel.
Egészen enyhe rezgés, például remegés vagy reszketés észlelésekor is felvillanhat.

A kijelzőn megjelenő szimbólumok (folytatás)
Szimbólumok

Funkció/jelentés

Javasolt teendő

A MEMÓRIÁBAN tárolt korábbi mérési eredmények
Átlagérték
TELJES ELEMKAPACITÁS
Az elemkapacitás-jelző mérés közbenALACSONY ELEMKAPACITÁS
A villogás alacsony elemkapacitást jelez
vagy

Ha a jel villog, cserélje ki az elemeket.

Mérés közbeni mozgás okozta instabil vérnyomásérték

Végezze el újra a mérést.
Mérés közben ne mozogjon.

A szisztolés és a diasztolés érték közötti különbség kevesebb, mint
10 mmHg
A nyomásérték felfújás közben nem emelkedettA mandzsetta nincs megfelelően felhelyezve
PULZUSKIJELZÉSI HIBA
Az eszköz nem megfelelően észleli a pulzust

Helyezze fel megfelelően a mandzsettát,és végezze el újra a mérést.

Vérnyomásmérő belső hibája

Vegye ki az elemeket, nyomja meg a
START gombot, majd helyezze beújból az elemeket. Ha a hiba továbbra isfennáll, lépjen kapcsolatba aforgalmazóval.

Magyar6

A vérnyomásmérő használata

1. Az elemek behelyezése/cseréje

1. Távolítsa el az elemtartó rekesz fedelét.
2. Vegye ki a használt elemeket az elemtartó rekeszből, és
helyezzen be újakat az ábrán látható módon. Ügyeljen arra,
hogy a (+) és a (–) pólus helyzete megfelelő legyen. Kizárólag
LR03 vagy AAA típusú elemeket használjon.
3. Helyezze vissza az elemtartó rekesz fedelét.

Elemtartó rekesz
fedeleHasznált elemekVIGYÁZATÚj elemekAz elemeket az elemtartó rekeszen feltüntetett módon
helyezze be, ellenkező esetben az eszköz nem fog működni.
Ha az(ALACSONY ELEMKAPACITÁS jele) villog akijelzőn, cserélje ki az elemeket. Ne használjon együtt
használt és új elemeket. Ez az elemek élettartalmának
csökkenését vagy az eszköz meghibásodását okozhatja.
Az(ALACSONY ELEMKAPACITÁS jele) nem jelenikmeg, ha az elemek lemerültek.
Az elemek élettartamát a környezeti hőmérséklet is befolyásolja. Alacsony hőmérsékleten az élettartamuk
rövidülhet. Átlagos esetben két új LR03/AAA elem körülbelül négy hónapig működteti az eszközt napi kétszeriméréssel számolva.
Csak a megadott elemtípusokat használja. Az eszközhöz mellékelt elemek a vérnyomásmérő kipróbálására
szolgálnak, élettartamuk korlátozott lehet.
Távolítsa el az elemeket, ha az eszközt hosszú ideig nem használja. Az elemek esetleges szivárgása
meghibásodást okozhat.
Magyar7

2. A mandzsetta felhelyezése
1. Helyezze a mandzsettát a csuklóra, körülbelül 1 centiméterrel a kézfej fölé, ajobb oldali ábrán látható módon.

Kb. 1 cm

2. Helyezze fel szorosan a mandzsettát a tépőzár használatával.
Megjegyzés: Pontos mérésekhez szorosan helyezze fel a mandzsettát, és a mérést
csupasz csuklón végezze.

3. Hogyan végezhetők pontos mérések?

A vérnyomás legpontosabb méréséhez:Mérés közben ne mozogjon és ne beszéljen.
Üljön kényelmesen. Helyezze a csuklóját az asztalra tenyérrel felfelé úgy, hogy a mandzsetta a szívével egy
vonalban legyen.
A mérés előtt lazítson 5–10 percig. Ha érzelmi stressz miatt izgatottnak vagy lehangoltnak érzi magát, az amérésben a normálisnál magasabb (vagy alacsonyabb) vérnyomásértékek formájában jelentkezik, a pulzus pedig
a normálisnál gyorsabb lesz.
Törekedjen arra, hogy minden nap nagyjából ugyanabban az időpontban végezze a mérést.
Az ember vérnyomása folyamatosan változik attól függően, hogy éppen mit csinál, mit evett vagy mit ivott. Ezek atényezők hirtelen és jelentős mértékű hatással vannak a vérnyomásra.
Mérés előtt közvetlenül ne végezzen testmozgást, és ne fürödjön. A mérés előtt pihenjen 20–30 percet.
Ne tegye keresztbe a lábát. A talpa legyen a talajon, és üljön egyenes háttal.
Az eszköz méréseinek alapja a szívritmus. Ha nagyon gyenge vagy szabálytalan a szívritmusa, az eszköz
nehézségekbe ütközhet a vérnyomás meghatározásakor.
Ha az eszköz rendellenes állapotot észlel, a mérés leáll, és az eszköz hibaszimbólumot jelenít meg.
A szimbólumok leírása a 6. oldalon található.
Az eszköz felnőttek általi használatra készült. Ha gyermekeken szeretné használni, először kérje kezelőorvosa
tanácsát. Az eszközt gyermekek felügyelet nélkül nem használhatják!
Magyar8

Az automata vérnyomásmérő működését az extrém hőmérséklet vagy páratartalom, illetve a tengerszint feletti
magasság is befolyásolhatja.

4. Mérés közben

Mérés közben a mandzsetta erősen szorítja a kart. Ez normális jelenség.

5. Mérés után

Ha az értékek kijelzésekor az eszköz kikapcsolásához megnyomja a START gombot, az eszköz a memóriában
tárolja az új értékeket.
Ha az értékek kijelzésekor az eszköz kikapcsolásához agombot nyomja meg, az eszköz nem tárolja az újértékeket.
Vegye le a mandzsettát, és rögzítse az adatokat.
Megjegyzés: Az eszköz automatikus leállítási funkcióval is rendelkezik, amely az aktuális adatokat eltárolja amemóriában, és a mérés után körülbelül egy perccel leállítja az eszközt. Ha ugyanazon a személyen
egymás után két mérést végez, hagyjon közöttük három perc szünetet.

Magyar9

Mérések
Megjegyzés:

Az UB-525 a használata esetén a felhasználó számára megfelelő mértékben fújja fel amandzsettát.

1. Helyezze fel a mandzsettát a csuklóra. A mandzsetta középvonala legyen aszívével egy magasságban, miközben üljön kényelmesen, és lazítson.
2. Nyomja meg a START gombot. A kijelző minden szegmense látható.
3. A nulla (0) érték röviden villog. A mérés megkezdésekor a kijelző megváltozik.
A mandzsetta elkezd felfúvódni. A mandzsetta erősen szorítja a kart.
Ez normális jelenség. A felfújás megkezdésekor a mérés automatikusan
elkezdődik, és a(szívdobogás szimbóluma) villog.
Megjegyzés:Ha bármikor le szeretné állítani a mandzsetta felfújását,nyomja meg újra a START gombot.

4. A mérés befejezése után az eszköz kijelzi a mért értékeket (megjelenik aszisztolés és a diasztolés vérnyomásérték, a pulzusszám, a WHO szerinti
osztályozás, illetve a szabálytalan szívritmus szimbóluma). A mandzsetta
automatikusan kiereszti a maradék levegőt, és teljesen leenged.

Mindenszegmens
Mindenszegmens
Nulla
érték
Nulla
értéka akijelzőn
kijelzőn
A felfújás megkezdődött
A felfújás
megkezdődött
Felfújás és mérés
Felfújásfolyamatban
és mérésfolyamatban

Szisztolés
vérnyomás
Szisztolés
vérnyomás
Diasztolésnyomás
nyomásDiasztolés
WHO szerinti
osztályozásWHOszerinti
Megjegyzés: Ha az új értékeket nem szeretné eltárolni a memóriában,Szabálytalan szívr.
szimbóluma
osztályozásaz értékek kijelzésekor nyomja meg agombot.
Szabálytalanszívr.
Pulzusszám
5. Az eszköz kikapcsolásához nyomja meg ismét a START gombot. Vegye
szimbóluma
le a mandzsettát.
P lá
Megjegyzés: Az eszköz automatikus leállítási funkcióval is rendelkezik.
Ha ugyanazon a személyen egymás után két mérést végez,hagyjon közöttük három perc szünetet.

Magyar10

A memóriában tárolt adatok megtekintése
Megjegyzés: Az eszköz az utolsó 60 mérést tárolja a memóriában.
Memória üzemmód
gombot.
1. Az eszköz kikapcsolásához nyomja meg a
Megjelenik az összes mérés átlaga és a kijelzett adatok száma. Ha
nincsenek adatok, a „0” érték jelenik meg. Az eszköz kikapcsolásához
Az összes adat átlaga
nyomja meg avagy a START gombot.
2. Az adatok (száma, illetve a mérési adatok) megjelenítéséhez használja az Az összes mérés száma
alábbi gombokat.
Az eszköz megjeleníti az összes mérés átlagát és a kijelzett adatok
számát.
Pl.: No.30
Legfrissebb adatAgomb ismételt megnyomásával az eszköz a következőket
jeleníti meg:Az összes mérés átlaga.
Adatok (száma, illetve a mért adatok).
Legrégebbi adat
Az eszköz a legfrissebb adattól kezdve sorrendben jeleníti meg
az adatokat.
Példa:
No.30 + adatok
No.29 + adatok
No.01 + adatok.
gombot, az
3. Ha a legrégebbi adat kijelzése után megnyomja aeszköz az 1. lépéshez ugrik, megjeleníti az összes mérés átlagát és akijelzett adatok számát.
4. Az eszköz kikapcsolásához nyomja meg a START gombot.
Egy percnyi tétlenség után az eszköz automatikusan kikapcsol.

A memóriában tárolt összes adat törlése

1. Nyomja meg és tartsa lenyomva agombot, amíg az eszköz automatikusan ki nem kapcsol.
Az eszköz ajelet jeleníti meg, törli a memóriában tárolt adatokat, miközben
jel villog, majd az eszköz automatikusan kikapcsol.
aMagyar 11

Kikapcsolás
gomb

gomb

gomb

gomb

Kikapcsolás

gombMit jelent a szabálytalan szívritmus/pitvarfibrilláció jelzése?
Ha mérés közben a vérnyomásmérő szabálytalan szívritmust észlel, a kijelzőn a mérési eredmények mellett
megjelenik a szabálytalan szívritmus/pitvarfibrilláció jelzése.
Megjegyzés: Ha gyakran jelenik meg a szabálytalan szívritmus/pitvarfibrilláció
keresse fel kezelőorvosát.

szimbóluma, javasoljuk, hogy

Mi az a pitvarfibrilláció?

A szívizom a szívben keletkező elektromos jel (ingerület) hatására összehúzódik, és vért pumpál a testbe. Akkor
fordul elő pitvarfibrilláció, ha a pitvarban az elektromos jel kaotikussá válik, és rendszertelen pulzust okoz.
A pitvarfibrilláció következtében vér marad a szívben, ami vérrögképződést és ezáltal szélütést vagy szívrohamot
okozhat.
A nyomás létrehozása

Vérnyomásjelző

A mérés során a nyomásjelző figyeli a vérnyomás változását.
A felfújás folyamatban van
Mérés folyamatban
A felfújás befejeződött

Magyar12

WHO szerinti osztályozás jelzője

A vérnyomásjelző mind a hat szakasza a WHO szerint meghatározott(a 14. oldalon ismertetett) vérnyomásosztályok egyikének felel meg.
Példa
Mérsékelt
hipertónia

Enyhe hipertónia

Magas-normális

WHO szerinti osztályozás jelzője
Súlyos hipertónia
Mérsékelt hipertónia
Enyhe hipertónia
Magas-normális
Normális
Optimális
: A jelző az aktuális értékek alapján
egy, a WHO szerinti osztályozásnak
megfelelő értéket mutat.

Mi az a vérnyomás?

Tudnivalók a vérnyomásról

A vérnyomás a vérnek az artériák falaira gyakorolt erőhatása. A szisztolés vérnyomás a szívizomösszehúzódásakor mérhető. A diasztolés nyomás a szívizom elernyedésekor mérhető. A vérnyomásmértékegysége a higanymilliméter (mmHg). A természetes vérnyomást a reggel, felkelés és evés előtt, mégnyugalmi helyzetben elvégzett vérnyomásmérés eredménye adja.

Mi az a hipertónia, és hogyan szabályozható?

A hipertónia rendellenesen magas vérnyomást jelent, amely kezelés nélkül számos egészségügyi problémát
okozhat, többek között szélütéshez vagy szívinfarktushoz vezethet. A hipertónia életmódváltással, a stressz
csökkentésével, illetve orvosi felügyelet mellett gyógyszeres úton kezelhető.
A hipertónia megelőzése vagy kontrollálása érdekében:Ne dohányozzon
Rendszeresen végezzen testedzést
Csökkentse a só- és zsírbevitelt
Rendszeresen járjon kivizsgálásraÜgyeljen rá, hogy ne legyen túlsúlyos
Magyar 13

Miért jó az otthoni vérnyomásmérés?

A kórházban vagy az orvosi rendelőben végzett vérnyomásmérés szorongást idézhet elő, ami az otthoni, nyugodt
körülményekhez képest akár 25–30 mmHg-rel magasabb értéket is eredményezhet. Az otthoni méréssel
csökkenthető a vérnyomásértékeket befolyásoló külső tényezők hatása, alátámasztható az orvos által végzettmérés, valamint pontosabb és teljesebb vérnyomáselőzmények állhatnak rendelkezésre.

A vérnyomásértékek osztályozása a WHO szerint

Az ember vérnyomása napi és szezonális rendszerességgel isnagymértékben változik. A nap folyamán a különféle körülményektől
függően az ingadozás mértéke akár 30–50 mmHg is lehet. A magas
vérnyomásban szenvedők esetében az ingadozás ennél nagyobb islehet. Normál esetben a vérnyomás munka vagy játék közben
emelkedik, a legalacsonyabb szintet pedig alvás közben éri el. Ezért
ne essen kétségbe egyetlen mérési eredménytől.
A normál vérnyomás meghatározásához a jelen útmutatóban
ismertetett eljárással végezzen méréseket lehetőleg minden nap
ugyanabban az időpontban. Rendszeres mérésekkel átfogóbb
vérnyomáselőzmények állíthatók össze. A vérnyomás
feljegyzésekor ne feledje az időt és a dátumot is rögzíteni.
A vérnyomásadatok értékeléséért forduljon kezelőorvosához.

Diasztolés vérnyomás

A vérnyomás változása

Harmadfokú hipertónia (súlyos)
Másodfokú hipertónia (mérsékelt)Elsőfokú hipertónia (enyhe)
Magas-normális
Normális
Optimális

Szisztolés vérnyomás
Jellemző napi ingadozás(öt percenként mérve)Szisztolés
Vérnyomás

Az Egészségügyi Világszervezet (WHO) meghatározta amagas vérnyomás minősítéséhez használt, kortól független,
szabványos értékeket. Ezek a jobb oldali ábrán láthatók.

Referencia: Journal ofHypertension 1999, Vol 17 No. 2

Diasztolésdu.

Magyar

14

Alvás
de.

Idő

Hibaelhárítás
Probléma
A kijelzőn nem jelenik
meg semmi akkor sem,ha az eszköz be vankapcsolva.
A mandzsetta nem
fúvódik fel.

Az eszköz nem mér.
Az értékek túl magasak
vagy túl alacsonyak.

Egyéb

Lehetséges ok
Az elemek lemerültek.

Javasolt teendő
Cserélje ki az elemeket újakra.
Helyezze be újra az elemeket úgy, hogy a pozitív és
negatív pólusok az elemtartó rekeszben jelzett helyen
legyenek.

Az elemek pólusai nincsenek megfelelő
helyzetben.

Az elemek kapacitása alacsony.
(ALACSONY ELEMKAPACITÁS
Az
Cserélje ki az elemeket újakra.
jele) villog. Ha az elemek teljesen
lemerültek, a jel nem jelenik meg.
A mandzsetta nincs megfelelően felhelyezve. Helyezze fel a mandzsettát megfelelően.
Mérés közben mozgatta a csuklóját, vagy
Mérés közben ne mozogjon és ne beszéljen.
megmozdult.
Üljön kényelmesen, és ne mozogjon.
A mandzsetta helyzete nem megfelelő.
Helyezze a karját az asztalra tenyérrel felfelé úgy,hogy a mandzsetta a szívével egy vonalban legyen.
Ha nagyon gyenge vagy szabálytalan a szívritmusa,az eszköz nehézségekbe ütközhet a vérnyomás
meghatározásakor.
Az érték eltér a kórházban vagy az orvosi
Olvassa el a „Miért jó az otthoni vérnyomásmérés?”rendelőben mérttől.
című szakaszt.
Vegye ki az elemeket. Helyezze vissza az elemeket
megfelelő helyzetben, majd próbálja meg újból
végrehajtani a mérést.

Megjegyzés: Ha a fent ismertetett módokon nem sikerül megoldani a problémát, lépjen kapcsolatba a forgalmazóval. Ne
próbálja felnyitni az eszközt, és ne próbálkozzon a javításával. Az ilyen kísérletek esetén a jótállás érvényét
veszti.

Magyar15

Karbantartás
Ne nyissa fel az eszközt. Az eszköz érzékeny elektronikai alkatrészeket és összetett levegőegységet tartalmaz,
amelyek ilyen esetben károsodhatnak. Ha a hibaelhárítási utasítások alapján nem tudja megoldani a problémát,lépjen kapcsolatba a helyi hivatalos forgalmazóval vagy az ügyfélszolgálati részlegünkkel. Az A&D ügyfélszolgálata
műszaki információkat, cserealkatrészeket és csereegységeket biztosít a hivatalos forgalmazók számára.
Az eszköz hosszú élettartamra készült. Ajánlott azonban az eszköz kétévenkénti vizsgálata a megfelelő működés
és pontosság érdekében. Karbantartásért lépjen kapcsolatba a helyi hivatalos forgalmazóval vagy az A&D
vállalattal.

Műszaki adatok
Típus
Mérési módszer
Mérési tartomány

Mérés pontossága
Tápellátás
Mérések száma
Csuklókerület
Osztályozás
Érintkező rész
Hasznos élettartam

UB-525
Oszcillometrikus mérés
Vérnyomás:
0 – 299 mmHg
Szisztolés vérnyomás: 60 – 279 mmHg
Diasztolés nyomás:
40 – 200 mmHg
Pulzus:40 – 180 szívverés/perc
Vérnyomás:±3 mmHg
Pulzus:±5%
2 db 1,5 V-os alkáli elem (LR03 vagy AAA)
Kb. 250 mérés AAA típusú alkáli elemek használata esetén, 170 mmHg
vérnyomásérték mellett szobahőmérsékleten (23 °C)
13,5–21,5 cm
Belső áramellátású orvosi berendezés (folyamatos működési mód)
BF típusú mandzsetta
Eszköz: 5 év (napi hat alkalommal való használat esetén)
Magyar16

Klinikai teszt
EMD
MemóriaMűködési feltételek
Szállítási/tárolási feltételek
Méretek
Tömeg
Behatolás elleni védettség

Az ISO81060-2：2013 előírásai szerint
A klinikai validációs vizsgálatban a K5 85 alanyra lett alkalmazva a diasztolés
vérnyomás megállapításához.
IEC 60601-1-2: 2014
Utolsó 60 mérés+10 °C – +40 °C / 15–85% relatív páratartalom / 800–1060 hPa-20 °C – +60 °C / 10–95% relatív páratartalom / 700–1060 hPa
Kb. 56 [Szé] x 88 [Ma] x 21,5 [Mé] mm
Kb. 90 g elemek nélkül
IP20

Megjegyzés: A műszaki adatok termékfejlesztési célból értesítés nélkül megváltozhatnak.
Az IP-besorolás a villamos gyártmányok burkolatai által nyújtott védettségi fokozat az IEC 60529
előírásai szerint. Az eszköz védett a 12 mm és annál nagyobb átmérőjű szilárd tárgyakkal (pl. az
ujjakkal) szemben. Az eszköz víz ellen nem védett.

Magyar17

MEMO

Magyar18Περιεχόμενα

Αγαπητοί πελάτες ................................................................................................................................................ 2Αρχικές σημειώσεις.............................................................................................................................................. 2Προφυλάξεις ........................................................................................................................................................ 2Αναγνώριση μερών .............................................................................................................................................. 4Σύμβολα .............................................................................................................................................................. 5Χρήση του πιεσόμετρου ...................................................................................................................................... 7
1. Τοποθέτηση/αντικατάσταση των μπαταριών ............................................................................................. 7
2. Εφαρμογή της μανσέτας ............................................................................................................................ 8
3. Λήψη ακριβών μετρήσεων ......................................................................................................................... 8
4. Μέτρηση ..................................................................................................................................................... 95. Μετά τη μέτρηση ........................................................................................................................................ 9Μετρήσεις .......................................................................................................................................................... 10Ανάκληση δεδομένων μνήμης ........................................................................................................................... 11Διαγραφή των αποθηκευμένων στη μνήμη δεδομένων ..................................................................................... 11Τι είναι η ένδειξη IHB/AFib; ................................................................................................................................ 12Τι σημαίνει AFib; ................................................................................................................................................ 12Ένδειξη γραμμής πίεσης ................................................................................................................................... 12Ένδειξη ταξινόμησης κατά ΠΟΥ ........................................................................................................................ 13Περί αρτηριακής πίεσης ..................................................................................................................................... 13Τι είναι η αρτηριακή πίεση; ........................................................................................................................... 13Τι είναι η υπέρταση και πώς ελέγχεται; ........................................................................................................ 13Γιατί είναι σημαντικό να μετράω την αρτηριακή πίεση στο σπίτι; ................................................................. 14Ταξινόμηση αρτηριακής πίεσης κατά ΠΟΥ................................................................................................... 14Διακυμάνσεις αρτηριακής πίεσης ................................................................................................................. 14Αντιμετώπιση προβλημάτων ............................................................................................................................. 15Συντήρηση ......................................................................................................................................................... 16Τεχνικά στοιχεία ................................................................................................................................................. 16Ελληνικά 1Αγαπητοί πελάτες
Συγχαρητήρια για την αγορά αυτού του πιεσόμετρου τελευταίας τεχνολογίας της A&D. Η συσκευή έχει σχεδιαστεί
για εύκολη χρήση και ακρίβεια, και θα σας διευκολύνει κατά την καθημερινή λήψη της αρτηριακής πίεσης.
Συνιστάται να διαβάσετε αυτό το εγχειρίδιο προσεκτικά πριν από την πρώτη χρήση της συσκευής.

Αρχικές σημειώσεις
Αυτή η συσκευή πληροί την ευρωπαϊκή οδηγία 93/42 ΕΟΚ περί ιατροτεχνολογικών προϊόντων. Η συμμόρφωση
υποδεικνύεται μέσω της σήμανσης συμμόρφωσης. (0123: Ο αριθμός αναφοράς του σχετικού
κοινοποιημένου οργανισμού)Η συσκευή έχει σχεδιαστεί για χρήση σε ενήλικες και όχι σε νεογέννητα βρέφη ή νήπια.
Περιβάλλον χρήσης. Η συσκευή προορίζεται για οικιακή χρήση από μη επαγγελματίες.
Αυτή η συσκευή έχει σχεδιαστεί για τη μέτρηση της αρτηριακής πίεσης και του καρδιακού ρυθμού για σκοπούς διάγνωσης.

Προφυλάξεις
Για την κατασκευή αυτής της συσκευής έχουν χρησιμοποιηθεί εξαρτήματα ακριβείας. Πρέπει να αποφεύγεται ηέκθεση σε ακραίες συνθήκες θερμοκρασίας, υγρασίας, άμεση ηλιακή ακτινοβολία, κρούσεις ή σκόνη.
Καθαρίστε τη συσκευή με ένα στεγνό, μαλακό πανί ή με ένα πανί εμποτισμένο με νερό και ουδέτερο
απορρυπαντικό. Μην χρησιμοποιείτε ποτέ οινόπνευμα, βενζίνη, διαλυτικό ή άλλες ισχυρές χημικές ουσίες για τονκαθαρισμό της συσκευής.
Μην σφίγγετε υπερβολικά τη μανσέτα για μεγάλα χρονικά διαστήματα, επειδή μπορεί να μειωθεί η διάρκεια ζωήςτων εξαρτημάτων.
Η συσκευή δεν είναι αδιάβροχη. Προστατέψτε τη συσκευή από βροχή, ιδρώτα και νερό.
Το αποτέλεσμα της μέτρησης ενδέχεται να αλλοιωθεί σε περίπτωση χρήσης της συσκευής κοντά σε τηλεοράσεις,φούρνους μικροκυμάτων, κινητά τηλέφωνα, ακτινογράφους ή άλλες συσκευές με ισχυρά ηλεκτρικά πεδία.
Οι παλιές συσκευές, τα εξαρτήματα και οι μπαταρίες τους δεν θεωρούνται οικιακά απορρίμματα και πρέπει νααπορρίπτονται σύμφωνα με τους ισχύοντες τοπικούς κανονισμούς.
Κάθε φορά που χρησιμοποιείτε τη συσκευή, διασφαλίζετε ότι είναι καθαρή.
Μην τροποποιείτε τη συσκευή. Ενδέχεται να προκληθούν ατυχήματα ή ζημιά στη συσκευή.
Για να μετρήσετε την αρτηριακή πίεση, ο καρπός πρέπει να πιέζεται επαρκώς από τη μανσέτα για προσωρινή
Ελληνικά 2διακοπή της ροής αίματος στις αρτηρίες. Αυτό μπορεί να προκαλέσει πόνο, μούδιασμα ή προσωρινό ερεθισμό τουκαρπού. Αυτή η κατάσταση προκύπτει ειδικά αν η μέτρηση επαναλαμβάνεται διαδοχικά. Ο πόνος, το μούδιασμα ή οερεθισμός υποχωρεί με την πάροδο του χρόνου.
Μια συσκευή ασύρματης επικοινωνίας, όπως οι συσκευές οικιακού δικτύου, τα κινητά τηλέφωνα, τα ασύρματα
τηλέφωνα και οι βάσεις τους, καθώς και τα γουόκι-τόκι, μπορεί να επηρεάσει το πιεσόμετρο. Επομένως, πρέπει ναδιατηρείται απόσταση τουλάχιστον 30 cm από τέτοιες συσκευές.
Η πολύ συχνή μέτρηση της αρτηριακής πίεσης μπορεί να προκαλέσει βλάβες λόγω της παρεμβολής στη ροή τουαίματος. Βεβαιωθείτε ότι η χρήση της συσκευής δεν εμποδίζει την κυκλοφορία του αίματος για μεγάλο χρονικό
διάστημα αν χρησιμοποιείτε τη συσκευή επανειλημμένα.
Δεν έχουν διεξαχθεί κλινικές δομικές σε νεογέννητα βρέφη και εγκύους. Μην χρησιμοποιείτε σε νεογέννητα βρέφη ή εγκύους.
Αν έχετε υποβληθεί σε μαστεκτομή, συμβουλευτείτε τον γιατρό σας πριν χρησιμοποιήσετε τη συσκευή.
Μην αφήνετε τα παιδιά να χρησιμοποιούν τη συσκευή χωρίς επιτήρηση και μην χρησιμοποιείτε τη συσκευή σεχώρο στον οποίο έχουν πρόσβαση βρέφη. Ενδέχεται να προκληθούν ατυχήματα ή ζημιές.
Η συσκευή περιέχει μικρά μέρη που εγκυμονούν κίνδυνο πνιγμού σε περίπτωση κατάποσης από βρέφη.
Μην αγγίζετε τις μπαταρίες και τον ασθενή ταυτόχρονα. Ενδέχεται να προκληθεί ηλεκτροπληξία.
Σε περίπτωση βλάβης μεμονωμένων εξαρτημάτων, το περίβλημα της εγγύς μανσέτας μπορεί να θερμανθεί
υπερβολικά και ενδεχομένως να προκαλέσει δυσλειτουργία.
Η χρήση παρελκομένων που δεν αναγράφονται σε αυτό το εγχειρίδιο μπορεί να επηρεάσει την ασφάλεια.
Σε περίπτωση βραχυκυκλώματος, η μπαταρία μπορεί να θερμανθεί υπερβολικά και να προκαλέσει, ενδεχομένως, εγκαύματα.
Αφήστε τη συσκευή να προσαρμοστεί στις συνθήκες του περιβάλλοντος πριν από τη χρήση (περίπου μία ώρα).
Μην φουσκώνετε τη μανσέτα χωρίς να την έχετε τυλίξει στον καρπό σας.

Αντενδείξεις

Ακολουθούν μερικές προφυλάξεις σχετικά με την κατάλληλη χρήση της συσκευής.
Μην τυλίγετε τη συσκευή σε καρπό που φέρει άλλες ιατρικές ηλεκτρικές συσκευές. Μπορεί να επηρεαστεί ηλειτουργία των συσκευών.
Τα άτομα με οξεία κυκλοφορική ανεπάρκεια στον βραχίονα θα πρέπει να συμβουλεύονται το γιατρό τους πρινχρησιμοποιήσουν τη συσκευή για την αποφυγή ιατρικών προβλημάτων.
Μην προβαίνετε σε διαγνώσεις με βάση τα αποτελέσματα της μέτρησης και μην ξεκινάτε μόνοι σας θεραπείες. Νασυμβουλεύεστε πάντα το γιατρό σας για την αξιολόγηση των αποτελεσμάτων και τη θεραπεία.
Μην τυλίγετε τη συσκευή σε καρπό με ανοιχτές πληγές.
Μην τυλίγετε τη συσκευή σε βραχίονα ο οποίος χρησιμοποιείται για ενδοφλέβια έγχυση φαρμάκων ή για μετάγγισηαίματος. Ενδέχεται να προκληθούν τραυματισμοί ή ατυχήματα.
Μην χρησιμοποιείτε τη συσκευή σε χώρους με εύφλεκτα αέρια, όπως π.χ. αναισθητικά αέρια. Μπορεί να προκληθεί έκρηξη.
Μην χρησιμοποιείτε τη συσκευή σε περιβάλλοντα με υψηλή συγκέντρωση οξυγόνου, όπως σε θαλάμους οξυγόνου
υψηλής πίεσης ή τέντες οξυγόνου. Ενδέχεται να προκληθεί πυρκαγιά ή έκρηξη.
Ελληνικά 3Αναγνώριση μερώνΟθόνηΚουμπί

(μέση τιμή, μνήμη και αλλαγή προκαθορισμένων παραμέτρων)START Κουμπί

Καπάκι θήκης μπαταριών

ΜανσέταΜΝΗΜΗΜέση τιμήΟθόνηΑλκαλικές μπαταρίες (LR03 ή AAA)Συστολική πίεση
Διαστολική πίεση

Ένδειξη ταξινόμησης κατά ΠΟΥκαιΈνδειξη γραμμής πίεσης
Σύμβολο IHB/AFibΣύμβολο καρδιάς

Καρδιακός ρυθμός

Ένδειξη μπαταρίας
Ελληνικά 4ΣύμβολαΣύμβολα τυπωμένα στο περίβλημα της συσκευής
ΣύμβολαΛειτουργία / Σημασία
Αναμονή και ενεργοποίηση της συσκευήςSYSΣυστολική αρτηριακή πίεση σε mmHg

DIA

Διαστολική αρτηριακή πίεση σε mmHg

PUL

Παλμοί ανά λεπτό
Οδηγός τοποθέτησης μπαταριών
Συνεχές ρεύμα

SN

Σειριακός αριθμός
Ημερομηνία κατασκευής
Τύπος BF: Η συσκευή και η μανσέτα έχουν σχεδιαστεί για την εξασφάλιση ειδικής προστασίας απόηλεκτροπληξία.
Ετικέτα ιατρικής συσκευής σύμφωνα με την οδηγία ΕΚIP

Διεθνές σύμβολο προστασίας
Ετικέτα ΑΗΗΕΚατασκευαστής
Αντιπρόσωπος στην ΕΕΑνατρέξτε στο εγχειρίδιο οδηγιών/φυλλάδιο
Διατηρείτε στεγνό

Ελληνικά 5Σύμβολα που εμφανίζονται στην οθόνη
ΣύμβολαΛειτουργία / Σημασία / Συνιστώμενη ενέργεια
Εμφανίζεται όταν βρίσκεται σε εξέλιξη μια μέτρηση. Αναβοσβήνει κατά την ανίχνευση παλμού.
Παραμείνετε κατά το δυνατόν ακίνητοι.
Σύμβολο IHB/AFib εμφανίζεται σε περίπτωση ανίχνευσης μη κανονικού καρδιακού παλμού.
Μπορεί να ανάψει αν ανιχνευτεί κάποια πολύ ελαφριά δόνηση, όπως π.χ. ρίγος ή τρέμουλο.

Σύμβολα που εμφανίζονται στην οθόνη (συνέχεια)ΣύμβολαΛειτουργία / Σημασία
Προηγούμενες μετρήσεις αποθηκευμένες στη ΜΝΗΜΗ
Μέσες τιμές
ΠΛΗΡΗΣ ΜΠΑΤΑΡΙΑΗ ένδειξη ισχύος της μπαταρίας κατά τη μέτρηση
ΧΑΜΗΛΗ ΜΠΑΤΑΡΙΑ
Η μπαταρία είναι χαμηλή όταν αναβοσβήνει

Ασταθής αρτηριακή πίεση λόγω κίνησης κατά τη μέτρησηήΣυνιστώμενη ενέργεια

Αντικαταστήστε όλες τις μπαταρίες μενέες όταν αναβοσβήνει η ένδειξη.
Επαναλάβετε τη μέτρηση.
Παραμείνετε ακίνητος κατά τηδιάρκεια της μέτρησης.

Η συστολική και διαστολή τιμή απέχουν μεταξύ τους πάνω από
10 mmHgΗ τιμή πίεσης δεν αυξήθηκε κατά το φούσκωμα
Εφαρμόστε τη μανσέτα σωστά καιεπαναλάβετε τη μέτρηση.
Η μανσέτα δεν έχει τοποθετηθεί σωστά
ΣΦΑΛΜΑ ΠΡΟΒΟΛΗΣ PUL.
Ο παλμός δεν ανιχνεύτηκε σωστά
Αφαιρέστε τις μπαταρίες και πατήστετο κουμπί START και, στη συνέχεια,τοποθετήστε ξανά τις μπαταρίες. ΑνΕσωτερικό σφάλμα πιεσόμετρου
το σφάλμα εξακολουθεί ναεμφανίζεται, επικοινωνήστε με τοναντιπρόσωπο.
Ελληνικά 6Χρήση του πιεσόμετρου

1. Τοποθέτηση/αντικατάσταση των μπαταριών
1. Αφαιρέστε το καπάκι της θήκης μπαταριών.
2. Αφαιρέστε τις χρησιμοποιημένες μπαταρίες από τη θήκημπαταριών και τοποθετήστε νέες σύμφωνα με την εικόνα,δίνοντας προσοχή στη σωστή πολικότητα (+ και -).
Χρησιμοποιείτε μόνο μπαταρίες LR03 ή AAA.
3. Τοποθετήστε το καπάκι της θήκης μπαταριών.

Καπάκι θήκης
μπαταριών

Χρησιμοποιημένες μπαταρίες

ΠΡΟΣΟΧΗΝέες μπαταρίες
Τοποθετήστε τις μπαταρίες σύμφωνα με την ένδειξη στη θήκημπαταριών. Αν δεν τοποθετηθούν σωστά, η συσκευή δεν θαλειτουργεί.
Αν το σύμβολο(ΧΑΜΗΛΗ ΜΠΑΤΑΡΙΑ) αναβοσβήνει
στην οθόνη, αντικαταστήστε όλες τις μπαταρίες με νέες. Μηναναμιγνύετε τις παλιές μπαταρίες με νέες. Ενδέχεται ναμειωθεί η διάρκεια ζωής των μπαταριών ή να προκληθεί
δυσλειτουργία της συσκευής.
Το σύμβολο (ΧΑΜΗΛΗ ΜΠΑΤΑΡΙΑ) δεν εμφανίζεται
όταν οι μπαταρίες είναι εξαντλημένες.
Η διάρκεια ζωής των μπαταριών διαφέρει ανάλογα με τη θερμοκρασία περιβάλλοντος και μπορεί να είναι μικρότερη
σε χαμηλές θερμοκρασίες. Γενικά, δύο νέες μπαταρίες LR03/AAA διαρκούν περίπου τέσσερις μήνες με δύομετρήσεις την ημέρα.
Χρησιμοποιείτε μόνο τους προβλεπόμενους τύπους μπαταριών. Οι μπαταρίες που παρέχονται με τη συσκευή
προορίζονται για τη δοκιμή της απόδοσης του πιεσόμετρου και ενδέχεται να έχουν περιορισμένη διάρκεια.
Αφαιρέστε τις μπαταρίες αν δεν σκοπεύετε να χρησιμοποιήσετε τη συσκευή για μεγάλο χρονικό διάστημα.
Ενδέχεται να προκύψει διαρροή των μπαταριών ή να προκληθεί δυσλειτουργία.
Ελληνικά 7

2. Εφαρμογή της μανσέτας
1. Τυλίξτε τη μανσέτα γύρω από τον βραχίονα περίπου 1 cm πάνω από το χέρι σαςσύμφωνα με την εικόνα στα δεξιά.

Περίπου 1 cm

2. Τοποθετήστε σφιχτά τη μανσέτα χρησιμοποιώντας την ταινία Velcro.
Σημείωση:Για ακριβείς μετρήσεις, τοποθετήστε σφιχτά τη μανσέτα και μετρήστε σεγυμνό καρπό.

3. Λήψη ακριβών μετρήσεων

Για ακριβή μέτρηση της αρτηριακής πίεσης:
Παραμείνετε ακίνητοι και μην μιλάτε κατά τη μέτρηση.
Καθίστε σε μια άνετη θέση. Τοποθετήστε τον αγκώνα σας σε ένα τραπέζι με την παλάμη προς τα πάνω και τημανσέτα στο επίπεδο της καρδιάς σας.
Χαλαρώστε για περίπου πέντε έως δέκα λεπτά πριν τη μέτρηση. Αν νιώθετε υπερένταση ή κατάθλιψη λόγωσυναισθηματικής φόρτισης, αυτή η φόρτιση θα αντικατοπτριστεί στη μέτρηση ως υψηλότερη (ή χαμηλότερη)αρτηριακή πίεση από το κανονικό και ο καρδιακός ρυθμός σας θα είναι ενδεχομένως γρηγορότερος από ότισυνήθως.
Προσπαθήστε να μετράτε την αρτηριακή πίεση την ίδια περίπου ώρα καθημερινά.
Η αρτηριακή πίεση ενός ατόμου παρουσιάζει συνεχώς διακυμάνσεις, ανάλογα με τη δραστηριότητά του, ενώ το
φαγητό και το ποτό που έχει καταναλώσει μπορεί να επιδράσουν σημαντικά και γρήγορα στην αρτηριακή πίεση.
Μην μετράτε την αρτηριακή πίεση αμέσως μετά από φυσική άσκηση ή μπάνιο. Ξεκουραστείτε για είκοσι έως τριάντα
λεπτά πριν τη μέτρηση.
Μην σταυρώνετε τα πόδια σας. Ακουμπήστε τα πέλματά σας στο δάπεδο και ισιώστε την πλάτη σας.
Αυτή η συσκευή βασίζει τις μετρήσεις στον καρδιακό παλμό. Αν έχετε πολύ αδύναμο ή ακανόνιστο καρδιακό παλμό,ενδέχεται η συσκευή να δυσκολεύεται να προσδιορίσει την αρτηριακή σας πίεση.
Αν η συσκευή ανιχνεύσει κάποια μη φυσιολογική κατάσταση, η μέτρηση διακόπτεται και εμφανίζεται ένα σύμβολοσφάλματος. Ανατρέξτε στη σελίδα 6 για την περιγραφή των συμβόλων.
Το πιεσόμετρο προορίζεται μόνο για χρήση από ενήλικες. Συμβουλευτείτε το γιατρό σας πριν τη χρήση αυτής τηςσυσκευής σε κάποιο παιδί. Τα παιδιά δεν πρέπει να χρησιμοποιούν τη συσκευή χωρίς επιτήρηση.
Η απόδοση του αυτόματου πιεσόμετρου μπορεί να επηρεαστεί από ακραίες θερμοκρασίες, υγρασία ή το υψόμετρο.
Ελληνικά 8

4. Μέτρηση

Κατά τη διάρκεια της μέτρησης είναι φυσιολογικό να αισθάνεστε ότι η μανσέτα σας σφίγγει.

5. Μετά τη μέτρηση

Κατά την εμφάνιση των τιμών, αν πατήσετε το κουμπί START για να απενεργοποιήσετε τη συσκευή, οι νέες τιμές
θα αποθηκευτούν στη μνήμη.
Κατά την εμφάνιση των τιμών, αν πατήσετε το κουμπί
για να απενεργοποιήσετε τη συσκευή, οι νέες τιμές δεν θααποθηκευτούν.
Αφαιρέστε τη μανσέτα και καταγράψτε τα δεδομένα σας.
Σημείωση:Η συσκευή παρέχεται με λειτουργία αυτόματης απενεργοποίησης, η οποία αποθηκεύει τα τρέχοντα
δεδομένα στη μνήμη και απενεργοποιεί αυτόματα τη συσκευή περίπου ένα λεπτό μετά τη μέτρηση.
Αφήστε να παρέλθει χρονικό διάστημα τουλάχιστον τριών λεπτών πριν την επανάληψη της μέτρησηςστο ίδιο άτομο.

Ελληνικά 9Μετρήσεις
Σημείωση:Μόλις χρησιμοποιηθεί το UB-525, θα παράσχει το κατάλληλο φούσκωμαγια τον χρήστη.

1. Τυλίξτε τη μανσέτα στον καρπό σας. Καθίστε άνετα με τη μανσέτα στο επίπεδοτης καρδιάς σας και χαλαρώστε.
2. Πατήστε το κουμπί START . Εμφανίζονται όλα τα τμήματα της οθόνης.
3. Η ένδειξη "μηδέν" (0) εμφανίζεται για λίγο και αναβοσβήνει. Στη συνέχεια, ηοθόνη αλλάζει μόλις ξεκινήσει η μέτρηση. Η μανσέτα αρχίζει να φουσκώνει.
Είναι φυσιολογικό να αισθάνεστε ότι η μανσέτα σας σφίγγει. Η μέτρηση ξεκινά
αυτόματα μόλις ξεκινήσει το φούσκωμα και η ένδειξη(σύμβολο καρδιάς)αναβοσβήνει.
Σημείωση:Αν θέλετε να διακόψετε το φούσκωμα ανά πάσα στιγμή,πατήστε ξανά το κουμπί START .

4. Μετά την ολοκλήρωση της μέτρησης, η συσκευή εμφανίζει τις τιμές (τιμές
συστολικής και διαστολικής πίεσης, καρδιακό ρυθμό, ταξινόμηση κατά ΠΟΥ,σύμβολο I.H.B.). Εκτονώνεται ο υπόλοιπος αέρας και η μανσέτα ξεφουσκώνει
αυτόματα τελείως.
Σημείωση:Αν δεν θέλετε να αποθηκεύσετε τις νέες τιμές στη μνήμη,πατήστε το κουμπί
κατά την παρουσίαση των τιμών.

5. Πατήστε ξανά το κουμπί START για να απενεργοποιήσετε τη συσκευή.
Αφαιρέστε τη μανσέτα.
Σημειώσεις: Η συσκευή παρέχεται με λειτουργία αυτόματης απενεργοποίησης.
Αφήστε να παρέλθει χρονικό διάστημα τουλάχιστον τριών λεπτών
πριν την επανάληψη της μέτρησης στο ίδιο άτομο.

Ελληνικά 10Όλα τα τμήματα
Ένδειξη "μηδέν"Το φούσκωμαξεκίνησεΦούσκωμα καιμέτρηση σε εξέλιξη
Συστολική πίεση
Διαστολική πίεση
Ταξινόμηση κατά ΠΟΥΣύμβολο I.H.B.
Καρδιακόςρυθμός
ρυθμός
Καρδιακός

Ανάκληση δεδομένων μνήμης

Απενεργοποίηση
Σημείωση: Η συσκευή αποθηκεύει τις τελευταίες 60 μετρήσεις στη μνήμη.
1. Πατήστε το κουμπί
κατά την απενεργοποίηση της συσκευής.
Λειτουργία μνήμης
Κουμπί
Εμφανίζονται η μέση τιμή όλων των μετρήσεων και ο αριθμός τωνδεδομένων. Αν δεν υπάρχουν δεδομένα, εμφανίζεται η ένδειξη "0".
Μέση τιμή όλων των δεδομένωνή START για να απενεργοποιήσετε τη συσκευή. Αριθμός όλων των δεδομένων
Πατήστε το κουμπί
2. Χρησιμοποιήστε τα παρακάτω κουμπιά για την εμφάνιση δεδομένων
(αριθμός δεδομένων και δεδομένα μέτρησης).
Κουμπί
Η συσκευή εμφανίζει τη μέση τιμή όλων των μετρήσεων και τοναριθμό των δεδομένων.
Π.χ.: Αρ. 30
Πιέζοντας κάθε φορά το κουμπί , η συσκευή εμφανίζει τα εξής:Πιο πρόσφατα δεδομένα
Μέσες τιμές των δεδομένων όλων των μετρήσεων που λήφθηκανστο μεσοδιάστημα.
Κουμπί
Δεδομένα (αριθμός δεδομένων και δεδομένα μέτρησης).
Τελευταία δεδομένα
Τα δεδομένα εμφανίζονται στη συσκευή ξεκινώντας από τα πιο πρόσφατα.
(παλιότερα)Αρ. 29 και δεδομένα
Στο παράδειγμα: Αρ. 30 και δεδομένα
Αρ. 01 και δεδομένα.
Κουμπί
3. Αν πατήσετε το κουμπί
μετά την εμφάνιση των παλαιότερωνδεδομένων, η συσκευή συνεχίζει στο βήμα 1 και εμφανίζονται η μέσητιμή όλων των μετρήσεων και ο αριθμός των δεδομένων.
4. Πατήστε ξανά το κουμπί START για να απενεργοποιήσετε τη συσκευή.
Απενεργοποίηση
Μετά από ένα λεπτό μη λειτουργίας, η συσκευή απενεργοποιείται αυτομάτως.

Διαγραφή των αποθηκευμένων στη μνήμη δεδομένων

1. Πατήστε παρατεταμένα το κουμπί

μέχρι να απενεργοποιηθεί αυτόματα η συσκευή.

Η συσκευή εμφανίζει το σύμβολο, διαγράφει τα δεδομένα που έχουν
και, στηαποθηκευτεί στη μνήμη ενώ παράλληλα αναβοσβήνει το σύμβολοσυνέχεια, απενεργοποιείται αυτόματα.
Ελληνικά 11Τι είναι η ένδειξη IHB/AFib;Όταν το πιεσόμετρο ανιχνεύει ακανόνιστο ρυθμό κατά τις μετρήσεις, εμφανίζεται στην οθόνη η ένδειξη IHB/AFib μετις τιμές μέτρησης.
Σημείωση: Συνιστάται να επικοινωνείτε με το γιατρό σας αν η ένδειξη

IHB/AFib εμφανίζεται συχνά.

Τι σημαίνει AFib;Η καρδιά συστέλλεται λόγω των ηλεκτρικών σημάτων και διοχετεύει το αίμα στο υπόλοιπο σώμα. Κολπική
μαρμαρυγή (AFib) προκύπτει όταν το ηλεκτρικό σήμα του καρδιακού κόλπου συγχέεται και διαταράσσει τονκαρδιακό παλμό. Η κολπική μαρμαρυγή (AFib) μπορεί να προκαλέσει στασιμότητα του αίματος στην καρδιά και,κατά συνέπεια, θρομβώσεις, οι οποίες μπορούν να οδηγήσουν σε εγκεφαλικό ή καρδιακή προσβολή.

Ένδειξη γραμμής πίεσης

Εφαρμογή πίεσηςΗ ένδειξη παρακολουθεί την πρόοδο της πίεσης κατά τημέτρηση.
Φούσκωμα σε εξέλιξη
Μέτρηση σε εξέλιξηΤο φούσκωμα ολοκληρώθηκε

Ελληνικά 12Ένδειξη ταξινόμησης κατά ΠΟΥΚάθε έξι τμήματα της ένδειξης γραμμής πίεσης αντιστοιχούν
στην ταξινόμηση αρτηριακής πίεσης κατά ΠΟΥ, η οποία
περιγράφεται στη σελίδα 14.
ΠαράδειγμαΜέτρια υπέρταση

Ήπια υπέρταση

Υψηλή κανονική

Ένδειξη ταξινόμησης κατά ΠΟΥΒαριά υπέρταση
Μέτρια υπέρταση
Ήπια υπέρταση
Υψηλή κανονική
Κανονική
Βέλτιστη: Η ένδειξη εμφανίζει ένα τμήμα,με βάση τα τρέχοντα δεδομένα,το οποίο αντιστοιχεί στην ταξινόμηση κατά ΠΟΥ.

Περί αρτηριακής πίεσηςΤι είναι η αρτηριακή πίεση;Η αρτηριακή πίεση είναι η δύναμη που ασκείται από το αίμα στα τοιχώματα των αρτηριών. Η συστολική πίεση είναιη πίεση που ασκείται κατά τη συστολή της καρδιάς. Η διαστολική πίεση είναι η πίεση που ασκείται κατά τη διαστολήτης καρδιάς. Η αρτηριακή πίεση μετράται σε χιλιοστόμετρα στήλης υδραργύρου (mmHg). Η φυσιολογική αρτηριακή
πίεση ενός ατόμου αντιστοιχεί στη βασική πίεση, η οποία μετράται το πρωί, όσο ακόμη το άτομο αναπαύεται καιπριν τη λήψη τροφής.

Τι είναι η υπέρταση και πώς ελέγχεται;Η υπέρταση, δηλαδή η αφύσικα υψηλή αρτηριακή πίεση, μπορεί να προκαλέσει πολλά προβλήματα υγείας αν δεναντιμετωπιστεί, συμπεριλαμβανομένων εγκεφαλικών και καρδιακών προσβολών. Η υπέρταση μπορεί να ελεγχθεί
αλλάζοντας τρόπο ζωής, αποφεύγοντας το άγχος και μέσω φαρμακευτικής αγωγής υπό την επίβλεψη γιατρού.
Για την πρόληψη ή τον έλεγχο της υπέρτασης:Μην καπνίζετε
Γυμνάζεστε τακτικά
Μειώστε την πρόσληψη αλατιού και λίπους
Ελέγχετε τακτικά την υγεία σαςΔιατηρείτε το βάρος σας σε φυσιολογικά επίπεδα
Ελληνικά 13Γιατί είναι σημαντικό να μετράω την αρτηριακή πίεση στο σπίτι;Ο Παγκόσμιος Οργανισμός Υγείας (ΠΟΥ) έχει ορίσει πρότυπα
αξιολόγησης της υψηλής αρτηριακής πίεσης, ανεξαρτήτως
ηλικίας, σύμφωνα με το διάγραμμα στα δεξιά.

Διακυμάνσεις αρτηριακής πίεσης

Η αρτηριακή πίεση ενός ατόμου παρουσιάζει σημαντικές
διακυμάνσεις σε καθημερινή και εποχιακή βάση. Η διακύμανση αυτήμπορεί να ανέρχεται σε 30 έως 50 mmHg λόγω των διαφορετικών
συνθηκών κατά τη διάρκεια της ημέρας. Στα άτομα που πάσχουναπό υπέρταση, η διακύμανση μπορεί να είναι ακόμη μεγαλύτερη.
Κανονικά, η αρτηριακή πίεση αυξάνεται κατά τη διάρκεια τηςεργασίας ή του παιχνιδιού και φτάνει στα χαμηλότερα επίπεδα κατάτη διάρκεια του ύπνου. Επομένως, τα αποτελέσματα μίας μέτρησης
δεν πρέπει να σας απασχολούν υπερβολικά.
Επαναλαμβάνετε τη μέτρηση την ίδια ώρα καθημερινά σύμφωνα μετη διαδικασία που περιγράφεται σε αυτό το εγχειρίδιο για ναδιαπιστώσετε τη φυσιολογική αρτηριακή σας πίεση. Μέσω τηςτακτικής μέτρησης θα έχετε ένα πιο ολοκληρωμένο ιστορικό
αρτηριακής πίεσης. Σημειώνετε την ημερομηνία και την ώρα κατάτην καταγραφή της αρτηριακής σας πίεσης. Συμβουλευτείτε τογιατρό σας για την ερμηνεία των δεδομένων αρτηριακής πίεσης.
Ελληνικά 14Υλικό αναφοράς: Journal ofHypertension 1999, τόμος 17 αρ. 2Υπέρταση βαθμού 3 (οξεία)Υπέρταση βαθμού 2 (μέτρια)Υπέρταση βαθμού 1 (ήπια)Υψηλή κανονική
Κανονική
ΒέλτιστηΣυστολική αρτηριακή πίεση
Τυπική διακύμανση στοπλαίσιο μιας ημέρας
(Μέτρηση ανά πέντε λεπτά)Αρτηριακή πίεση

Ταξινόμηση αρτηριακής πίεσης κατά ΠΟΥΔιαστολική αρτηριακή πίεση

Κατά τη μέτρηση της αρτηριακής πίεσης σε κάποια κλινική ή ιατρείο μπορεί να προκληθεί αναστάτωση, μεαποτέλεσμα να προκύψει υψηλότερη τιμή κατά 25 έως 30 mmHg σε σχέση με τη μέτρηση στο σπίτι. Κατά τημέτρηση στο σπίτι περιορίζονται οι επιπτώσεις των εξωτερικών επιδράσεων στην τιμή της αρτηριακής πίεσης, ενώοι τιμές που προκύπτουν μπορούν να χρησιμοποιηθούν υποστηρικτικά προς τις τιμές που λαμβάνονται από τογιατρό για ένα πιο ακριβές και πλήρες ιστορικό αρτηριακής πίεσης.

ΣυστολικήΎπνοςΔιαστολική
ΜΜ

ΠΜ

Ώρα

Αντιμετώπιση προβλημάτων
ΠρόβλημαΔεν εμφανίζεται τίποτα
στην οθόνη, ακόμη καιόταν η συσκευή είναι
ενεργοποιημένη.
Η μανσέτα δενφουσκώνει.

Η συσκευή δεν μετράει.
Οι τιμές είναι πολύυψηλές ή πολύ χαμηλές.

ΆλλοΣημείωση:Πιθανή αιτία
Οι μπαταρίες έχουν εξαντληθεί.

Συνιστώμενη ενέργεια
Αντικαταστήστε όλες τις μπαταρίες με νέες.
Τοποθετήστε ξανά τις μπαταρίες με τονΟι μπαταρίες δεν έχουν τοποθετηθεί μεαρνητικό και τον θετικό πόλο να αντιστοιχούν
τη σωστή πολικότητα.
στις ενδείξεις της θήκης μπαταριών.
Η τάση των μπαταριών είναι πολύΤο σύμβολο (ΧΑΜΗΛΗχαμηλή.
Αντικαταστήστε όλες τις μπαταρίες με νέες.
ΜΠΑΤΑΡΙΑ) αναβοσβήνει. Αν οι
μπαταρίες έχουν εξαντληθεί πλήρως, τοσύμβολο δεν εμφανίζεται.
Η μανσέτα δεν έχει εφαρμόσει σωστά.
Εφαρμόστε τη μανσέτα σωστά.
Μετακινήσατε τον καρπό σας ήΜην κινήστε και μην μιλάτε κατά τη διάρκεια τηςκουνηθήκατε κατά τη μέτρηση.
μέτρησης.
Καθίστε άνετα και μην κινήστε.
Τοποθετήστε το βραχίονά σας σε ένα τραπέζιΗ θέση της μανσέτας δεν είναι σωστή.
με την παλάμη προς τα πάνω και τη μανσέταστο επίπεδο της καρδιάς σας.
Αν έχετε πολύ αδύναμο ή ακανόνιστο καρδιακό
παλμό, ενδέχεται η συσκευή να δυσκολεύεται
να προσδιορίσει την αρτηριακή σας πίεση.
Η τιμή διαφέρει από την τιμή μέτρησης Ανατρέξτε στην ενότητα "Γιατί είναι σημαντικό
στην κλινική ή στο ιατρείο.
να μετράω την αρτηριακή πίεση στο σπίτι;".
Αφαιρέστε τις μπαταρίες. Τοποθετήστε τις ξανάσωστά και επαναλάβετε τη μέτρηση.

Αν το πρόβλημα δεν λυθεί μέσω των ενεργειών που περιγράφονται παραπάνω, επικοινωνήστε με τοναντιπρόσωπο. Μην επιχειρείτε να ανοίξετε ή να επισκευάσετε το προϊόν οι ίδιοι, επειδή κάθε τέτοια
προσπάθεια ακυρώνει την εγγύηση.
Ελληνικά 15Συντήρηση

Μην ανοίγετε τη συσκευή. Έχει ευαίσθητα ηλεκτρικά εξαρτήματα και μια περίπλοκη μονάδα αέρα που μπορεί ναυποστεί βλάβη. Αν δεν μπορείτε να επιδιορθώσετε το πρόβλημα σύμφωνα με τις οδηγίες αντιμετώπισης
προβλημάτων, επικοινωνήστε με τον εξουσιοδοτημένο αντιπρόσωπο της περιοχής σας ή το τμήμα εξυπηρέτησης
πελατών. Το τμήμα εξυπηρέτησης πελατών της A&D παρέχει τεχνικές πληροφορίες, ανταλλακτικά και μονάδες σεεξουσιοδοτημένους αντιπροσώπους.
Η συσκευή έχει σχεδιαστεί και κατασκευαστεί για μεγάλη διάρκεια ζωής. Ωστόσο, συνιστάται γενικά να επιθεωρείτε
τη συσκευή ανά 2 έτη, ώστε να διασφαλίζετε την κατάλληλη λειτουργία και την ακρίβειά της. Επικοινωνήστε με τονεξουσιοδοτημένο αντιπρόσωπο της περιοχής σας ή με την A&D για συντήρηση.

Τεχνικά στοιχεία
ΤύποςΜέθοδος μέτρησης
Εύρος μέτρησης

Ακρίβεια μέτρησης
Τροφοδοσία
Αριθμός μετρήσεων
Περίμετρος καρπού
Ταξινόμηση
Εφαρμοζόμενο μέρος

UB-525Ταλαντωσιμετρική μέτρηση
Πίεση:
0 – 299 mmHg
Συστολική πίεση:60 – 279 mmHg
Διαστολική πίεση:40 – 200 mmHg
Καρδιακός παλμός:40 – 180 παλμοί/λεπτό
Πίεση:±3 mmHgΚαρδιακός ρυθμός: ±5 %2 αλκαλικές μπαταρίες 1,5 V (LR03 ή AAA)
250 περίπου μετρήσεις, όταν χρησιμοποιούνται αλκαλικές μπαταρίες ΑΑΑ, με τιμήπίεσης 170 mmHg σε θερμοκρασία δωματίου 23 °C.
13,5 - 21,5 cm
Ηλεκτρικός ιατρικός εξοπλισμός εσωτερικής τροφοδοσίας (Κατάσταση συνεχούς
λειτουργίας)Μανσέτα τύπου BF
Ελληνικά 16Ωφέλιμη διάρκεια ζωήςΚλινική δοκιμή

Συσκευή: 5 έτη (με χρήση έξι φορές την ημέρα)Κατά ISO81060-2：2013Στην μελέτη κλινικής επικύρωσης, το K5 χρησιμοποιήθηκε σε 85 ασθενείς για τονπροσδιορισμό της διαστολικής αρτηριακής πίεσης.
IEC 60601-1-2: 2014
Τελευταίες 60 μετρήσεις
+10 έως +40 °C / 15 έως 85% σχετ. υγρασία / 800 έως 1.060 hPa

EMD
Μνήμη
Συνθήκες λειτουργίας
Συνθήκες
-20 έως +60 °C / 10 έως 95% σχετ. υγρασία / 700 έως 1.060 hPaμεταφοράς/αποθήκευσης
Διαστάσεις
Περίπου 56 [Π] x 88 [Υ] x 21,5 [Β] mm
ΒάροςΠερίπου 90 g, χωρίς τις μπαταρίες
Προστασία από εισχώρηση IP20Σημείωση:Οι προδιαγραφές υπόκεινται σε αλλαγή για βελτίωση χωρίς προηγούμενη ειδοποίηση.
Η ταξινόμηση IP αφορά το βαθμό προστασίας που παρέχεται από τα περιβλήματα κατά IEC 60529. Αυτήη συσκευή προστατεύεται από ξένα στερεά αντικείμενα διαμέτρου 12 mm και πάνω, όπως τα δάχτυλα.
Η συσκευή δεν προστατεύεται από την εισχώρηση νερού.

Ελληνικά 17ΥΠΟΜΝΗΜΑΕλληνικά 18

Sadržaj

Dragi korisnici ...................................................................................................................................................... 2
Uvodne napomene .............................................................................................................................................. 2
Mjera opreza ........................................................................................................................................................ 2
Identifikacija dijelova ............................................................................................................................................ 4
Simboli ................................................................................................................................................................. 5
Uporaba tlakomjera ............................................................................................................................................. 7
1. Stavljanje/mijenjanje baterija ...................................................................................................................... 7
2. Stavljanje manšete ..................................................................................................................................... 8
3. Kako dobiti precizna mjerenja .................................................................................................................... 8
4. Mjerenje ..................................................................................................................................................... 9
5. Nakon mjerenja .......................................................................................................................................... 9
Mjerenja ............................................................................................................................................................. 10
Dohvaćanje podataka iz memorije ..................................................................................................................... 11
Brisanje podataka pohranjenih u memoriji ......................................................................................................... 11
Što je pokazivač IHB/AFib? ............................................................................................................................... 12
Što je AFib? ....................................................................................................................................................... 12
Pokazatelj trake tlaka ......................................................................................................................................... 12
Pokazatelj klasifikacije WHO-a .......................................................................................................................... 13
O krvnom tlaku................................................................................................................................................... 13
Što je krvni tlak? ........................................................................................................................................... 13
Što je hipertenzija i kako se kontrolira? ........................................................................................................ 13
Zašto mjeriti krvni tlak kod kuće? ................................................................................................................. 14
Klasifikacija krvnog tlaka prema WHO-u ...................................................................................................... 14
Varijacije krvnog tlaka .................................................................................................................................. 14
Rješavanje problema ......................................................................................................................................... 15Održavanje ........................................................................................................................................................ 16
Tehnički podatci ................................................................................................................................................. 16

Hrvatski 1

Dragi korisniciČestitamo na kupnji najmodernijeg tlakomjera poduzeća A&D. Ovaj je uređaj napravljen radi lakoće uporabe i
preciznosti i zahvaljujući njemu olakšat ćete si dnevni režim mjerenja krvnog tlaka.
Preporučujemo vam da pažljivo pročitate ovaj priručnik prije nego što prvi put upotrijebite uređaj.

Uvodne napomene
Ovaj uređaj u skladu je s europskom direktivom 93/42/EEZ za medicinske proizvode. Dokaz je tomu znak
. (0123: referentni broj uključenom prijavljenom tijelu).
usklađenostiUređaj je namijenjen uporabi na odraslima, ne na novorođenčadi ili malim bebama.
Okolina za uporabu. Predviđeno je da uređaj rabite sami u kućnoj zdravstvenoj okolini.
Svrha je ovoga uređaja mjerenje krvnog tlaka i brzine pulsa osoba radi dijagnoze.

Mjera opreza

U izgradnji ovoga uređaja upotrijebljeni su dijelovi kojima se jamči preciznost. Treba izbjegavati ekstremne
temperature, vlagu, izravnu Sunčevu svjetlost, udarce ili prašinu.
Očistite uređaj suhom i mekanom krpom ili krpom natopljenom vodom i neutralnim deterdžentom. Nikada ne
upotrebljavajte alkohol, benzen, razrjeđivač ili druge teške kemikalije za čišćenje uređaja.
Nemojte tijesno savijati manšetu na dulja razdoblja, budući da time možete smanjiti radni vijek dijelova.
Uređaj nije vodootporan. Spriječite prljanje uređaja kišom, znojem i vodom.
Mjerenja mogu biti iskrivljena ako se uređaj rabi blizu televizora, mikrovalnih pećnica, mobilnih telefona, rendgena
ili drugih uređaja koji imaju jaka električna polja.
Upotrijebljena oprema, dijelovi i baterije ne smatraju se običnim kućanskim otpadom i moraju se zbrinuti u skladu s
mjerodavnim lokalnim propisima.
Kada ponovo upotrebljavate uređaj, provjerite je li čist.
Ne preinačavajte uređaj. Time možete izazvati nesreće ili oštećenja uređaja.
Da biste izmjerili krvni tlak, zapešće mora biti stisnuto manšetom dovoljno jako da se privremeno zaustavi protok
krvi kroz arteriju. Time možete prouzročiti bol, utrnulost ili privremeno crvenilo zapešća. Ovo se pojavljuje osobito
kada se mjerenje ponavlja jedno za drugim. Bilo koja bol, utrnulost ili crvenilo s vremenom nestaju.
Uređaji za bežičnu komunikaciju, primjerice kućni uređaji za umrežavanje, mobilni telefoni, bežični telefoni i njihove bazne
Hrvatski 2

stanice i voki-tokiji mogu utjecati na ovaj tlakomjer. Stoga tlakomjer od tih uređaja treba biti udaljen najmanje 30 cm.
Ako prečesto mjerite krvni tlak, možete prouzročiti štetu zbog ometanja protoka krvi. Pobrinite se da u slučaju
ponavljane uporabe uređaja ne prouzročite produljeno narušavanje cirkulacije.
Klinička testiranja nisu izvedena na novorođenoj djeci i na trudnicama. Ne upotrebljavajte na novorođenoj djeci i
trudnicama.
Ako ste imali mastektomiju, posavjetujte se s liječnikom prije uporabe uređaja.
Nemojte dopustiti djeci da sama rabe uređaj i ne upotrebljavajte uređaj na mjestu koje je na dohvat maloj djeci.
Time možete izazvati nesreće ili štetu.
Postoje mali dijelovi zbog kojih može nastati rizik od gušenja ako ih mala djeca zabunom progutaju.
Ne dodirujte baterije i pacijenta u isto vrijeme. Time možete prouzročiti strujni udar.
U slučaju kvara pojedinačnih dijelova kućište blizu manšete može se zagrijati te potencijalno prouzročiti poremećaj u radu.
Ako upotrebljavate dodatnu opremu koja nije navedena u ovom priručniku, možete ugroziti sigurnost.
Ako nastane kratki spoj baterije, može se zagrijati i biti potencijalnim uzrokom opeklina.
Neka se uređaj prilagodi na okolno okruženje prije uporabe (otprilike jedan sat).
Nemojte napuhavati prije nego što omotate manšetu oko zapešća.

Kontraindikacije

U nastavku su navedene mjere opreza za pravilnu uporabu uređaja.
Ne stavljajte uređaj na zapešće ako je na njega prikačena neka druga medicinska električna oprema. Opremamožda neće ispravno raditi.
Ljudi koji imaju težak manjak cirkulacije u ruci moraju se posavjetovati s liječnikom prije uporabe uređaja da bi
spriječili medicinske probleme.
Nemojte sami postavljati dijagnozu nakon mjernih rezultata i sami počinjati liječenje. Uvijek se posavjetujte sliječnikom radi vrednovanja rezultata i radi liječenja.
Ne stavljajte uređaj na zapešće na kojem je nezacijeljena rana.
Ne stavljajte uređaj na ruku koja prima intravenski drip ili transfuziju krvi. Time možete izazvati ozljede ili nesreće.
Ne upotrebljavajte uređaj u prostoru gdje ima zapaljivih plinova, primjerice anestetičkih plinova. Možete izazvati
eksploziju.
Ne upotrebljavajte uređaj u okolinama visoke koncentracije kisika, primjerice u visokotlačnoj komori za kisik ilišatoru s kisikom. Možete prouzročiti požar ili eksploziju.

Hrvatski 3

Identifikacija dijelova
Zaslongumb (prosjek, memorija i promjena unaprijed postavljenih parametara)

START Gumb
Poklopac baterije

ManšetaAlkalne baterije (LR03 ili AAA)

ZaslonMEMORIJA
Prosječno

Sistolički tlak
Dijastolički tlak

Pokazatelj klasifikacije WHO-ai
Pokazatelj trake tlaka
Simbol IHB/AFibOznaka srca

Brzina pulsa

Pokazivač baterije

Hrvatski 4

Simboli

Simboli koji su otisnuti na kućištu uređaja
Simboli

Funkcija / značenje
Standby i Uključite uređaj

SYS

Sistolički krvni tlak u mmHg

DIA

Dijastolički krvni tlak u mmHg

PUL

Puls po minuti
Vodič za ugradnju baterije
Istosmjerna struja

SN

Serijski broj
Datum proizvodnje
Tip BF: uređaj i manšeta napravljeni su radi posebne zaštite od strujnih udara
Naljepnica medicinskog uređaja prema direktivi EK-a

IP

Simbol međunarodne zaštite
Naljepnica WEEE (OEEO)
Proizvođač
Predstavnik u EU-u
Pogledajte priručnik s uputama / knjižicu
Neka bude suho

Hrvatski 5

Simboli koji se pojavljuju na zaslonu
Simboli

Funkcija / značenje / preporučena radnja
Pojavljuje se dok je mjerenje u tijeku. Treperi kada je otkriven puls.
Budite što je moguće mirniji.
Simbol IHB/AFib pojavljuje se kada se otkriju nepravilni otkucaji srca.
Može zasvijetliti kada se otkrije vrlo lagana vibracija, primjerice drhtanje ili tresenje.

Simboli koji se pojavljuju na zaslonu (nastavak)
Simboli

Funkcija / značenje
Prijašnja mjerenja pohranjena u MEMORIJI
Prosječni podatci
PUNA BATERIJA
Pokazivač napunjenosti baterije tijekom mjerenja
BATERIJA PRI KRAJU
Baterija je pri kraju kada treperi

Nestabilan krvni tlak zbog pokreta tijekom mjerenja
ili

Preporučena radnja

Zamijenite sve baterije novima kadapokazivač treperi.
Ponovo izvedite mjerenje.
Budite mirni tijekom mjerenja.

Sistolička i dijastolička vrijednost u međusobnom su rasponu do
10 mmHg.
Vrijednost tlaka nije se povećala tijekom napuhavanja.
Pravilno stavite manšetu i ponovno
izvedite mjerenje.
Manšeta nije stavljena na pravilan način.
PUL.DISPLAY ERROR
Puls nije pravilno očitan.
Izvadite baterije i pritisnite gumb
START , a zatim ponovno stavite
Interna pogreška tlakomjera
baterije. Ako se pogreška i dalje
pojavljuje, obratite se prodavaču.
Hrvatski 6

1. Stavljanje/mijenjanje baterija

Uporaba tlakomjera

1. Uklonite poklopac baterije.
2. Izvadite iskorištene baterije i umetnite nove baterije u odjeljak
za bateriju kako je prikazano i pripazite da su polariteti (+ i –)točno stavljeni. Upotrebljavajte samo baterije tipa LR03 ili AAA.

Poklopac baterije

3. Stavite poklopac baterije.
Iskorištene baterije

UPOZORENJA

Umetnite baterije kako je prikazano u odjeljak za bateriju.
Nove baterije
Ako ih nepravilno stavite, uređaj neće raditi.
Kada na zaslonu treperi(oznaka LOW BATTERY(BATERIJA PRI KRAJU)), zamijenite sve baterije novima. Ne
upotrebljavajte istodobno stare i nove baterije. Tako možete
skratiti radni vijek baterije ili prouzročiti poremećaj u raduuređaja.
(oznaka LOW BATTERY (BATERIJA PRI KRAJU)) ne
pojavljuje se kada se baterije isprazne.
Radni vijek baterije varira ovisno o temperaturi okoline i može biti kraći na niskim temperaturama. Općenito, dvije
nove baterije LR03/AAA traju otprilike četiri mjeseca kada se rabe dvaput za mjerenje svaki dan.
Upotrebljavajte samo navedene baterije. Baterije koje se isporučuju s uređajem namijenjene su ispitivanju izvedbe
tlakomjera i možda imaju ograničen radni vijek.
Izvadite baterije ako se uređaj neće rabiti dulje vrijeme. Baterije bi mogle procuriti i prouzročiti poremećaj u radu.
Hrvatski 7

2. Stavljanje manšete
1. Omotajte manšetu oko zapešća otprilike 1 cm iznad ruke kako je prikazano na
slici desno.

Otprilike 1 cm

2. Dobro stegnite manšetu uporabom trake Velcro.
Napomena: da biste dobili precizno mjerenje, dobro stegnite manšetu i mjerite na
golom zapešću.

3. Kako dobiti precizna mjerenja

Da biste dobili najpreciznije mjerenje krvnog tlaka:
Budite mirni i tihi tijekom mjerenja.
Sjednite u udoban položaj. Stavite lakat na stol s dlanom okrenutim prema gore, a manšeta treba biti u razini srca.
Opustite se pet do deset minuta prije mjerenja. Ako ste uzbuđeni ili utučeni zbog emocionalnog stresa, taj će sestres odraziti u višem (ili nižem) očitanju krvnog tlaka od normalnog, a puls će obično biti brži od normalnog.
Pokušajte mjeriti krvni tlak otprilike u isto vrijeme svaki dan.
Krvni tlak pojedinca stalno varira, ovisno o onome što radi, što je pojeo i što pije, što sve može vrlo jako i brzo
utjecati na krvni tlak.
Ne mjerite odmah nakon vježbanja ili kupke. Odmorite se dvadeset ili trideset minuta prije mjerenja.
Nemojte prekrižiti noge. Neka vam stopala budu paralelna s podom i ispravite leđa.
Kod ovoga uređaja mjerenja se temelje na otkucajima srca. Ako imate vrlo slabe ili nepravilne otkucaje srca, možda
nastanu poteškoće za uređaj pri određivanju krvnog tlaka.
Ako uređaj otkrije abnormalno stanje, mjerenje se zaustavlja i prikazuje se simbol pogreške. Pogledajte 6 stranicu
radi opisa simbola.
Predviđeno je da ovaj tlakomjer rabe odrasli. Posavjetujte se s liječnikom prije nego što upotrijebite ovaj uređaj na
djetetu. Dijete ne bi smjelo upotrebljavati ovaj uređaj bez nadzora.
Na rad automatskog tlakomjera mogu utjecati prekomjerne temperature ili vlaga ili visina.
Hrvatski 8

4. Mjerenje

Normalno je da vas tijekom mjerenja manšeta jako stegne.

5. Nakon mjerenja

Dok su prikazana očitanja, ako pritisnete gumb START da biste isključili uređaj, nova se očitanja pohranjuju umemoriju.
Dok su prikazana očitanja, ako pritisnete gumbda biste isključili uređaj, nova se očitanja ne pohranjuju.
Uklonite manšetu i zabilježite svoje podatke.
Napomene: uređaj ima funkciju automatskog isključivanja napajanja kojom se trenutačni podatci pohranjuju umemoriju, a uređaj se gasi automatski minutu nakon mjerenja. Neka prođu barem tri minute između
mjerenja na istoj osobi.

Hrvatski 9

Mjerenja
Napomena:kada se dio UB-525 jednom upotrijebi, omogućava odgovarajuće
napuhavanje za korisnika.

1. Omotajte manšetu oko zapešća. Sjednite u udoban položaj i postavite manšetu
na istu razinu na kojoj vam je i srce i opustite se.
2. Pritisnite gumb START . Prikazuju se svi odjeljci zaslona.
3. Prikazuje se nula (0) i kratko treperi. Tada se mijenja prikaz na zaslonu kako
mjerenje počinje. Manšeta se počinje napuhavati. Normalno je da vas manšeta
jako stegne. Mjerenje počinje automatski kada počne napuhavanje i(oznaka srca) treperi.

Svi odjeljci

Prikazuje se nula
Počelo je napuhavanje

Napomena: ako u bilo kojem trenutku želite zaustaviti napuhavanje, ponovo
pritisnite gumb START .

U tijeku su napuhavanje
i mjerenje
4. Kada je mjerenje završeno, na uređaju se prikazuju očitanja (očitanja sistoličkog

i dijastoličkog tlaka, brzine pulsa, klasifikacije WHO-a, simbola I.H.B.). Iz
manšete se ispušta preostali zrak i automatski se potpuno ispuhuje.
Napomena: ako ne želite pohraniti nova očitanja u memoriju, pritisnite gumbdok su prikazana očitanja.
5. Ponovo pritisnite gumb START da biste isključili uređaj. Uklonite manšetu.
Napomene: uređaj ima funkciju automatskog isključivanja napajanja.
Neka prođu barem tri minute između mjerenja na istoj osobi.

Hrvatski 10

Sistolički tlak
Dijastolički tlak
Klasifikacija WHO-a
Simbol I.H.B.
Brzina pulsa

Dohvaćanje podataka iz memorije
Napomena: u memoriji uređaja pohranjuje se zadnjih 60 mjerenja.
Način rada memorije
prilikom isključivanja uređaj.
1. Pritisnite gumbPrikazuju se prosjek svih mjerenja i broj podataka. Ako nema podataka,
Prosjek svih podataka
prikazuje se »0«. Pritisnite gumb ili START da biste isključili uređaj.
Broj svih podataka
2. Upotrijebite sljedeće gumbe za prikaz podataka (broja i mjernih podataka).
Na uređaju se prikazuje prosjek svih mjerenja i broj podataka.
Ako svaki put pritisnete gumb , na uređaju se prikazuje sljedeće:Npr. br. 30
 Prosječni podatci svih mjerenja u međuvremenu.
Najnoviji podatci
 Podatci (broja i mjernih podataka).
Na uređaju se prikazuju podatci počevši od najnovijih podataka.
U primjeru: br. 30 i podatci
br. 29 i podatci
br. 01 i podatci.
Posljednji podatci(najstariji)nakon prikaza najstarijih podataka, uređaj ide
3. Ako pritisnete gumbna 1. korak i prikazuje se prosjek svih mjerenja i broj podataka.
4. Pritisnite gumb START da biste isključili uređaj.
Nakon minute neaktivnosti, uređaj se automatski gasi.

Isključite
gumb

gumb

gumb

gumb

Isključite

Brisanje podataka pohranjenih u memoriji
1. Pritisnite i držite pritisnutim gumbdok se uređaj ne ugasi automatski.

Na uređaju se prikazuje oznaka, brišu se podatci pohranjeni u memoriji dok
treperi oznaka
i automatski se gasi.

Hrvatski 11Što je pokazivač IHB/AFib?
Kada tlakomjer otkrije nepravilan ritam tijekom mjerenja, na zaslonu se prikazuje pokazivač IHB/AFib s mjernim
vrijednostima.
Napomena: preporučujemo da se obratite liječniku ako često vidite ovaj

pokazivač IHB/AFib.

Što je AFib?

Srce se steže zbog električnih signala koji nastaju u srcu i šalje krv po tijelu. Atrijska fibrilacija (AFib) nastaje kadaelektrični signal u atriju postane nepravilan čime dolazi do smetnji u intervalu pulsa. AFib može biti uzrokom
zaostajanja krvi u srcu, čime lako mogu nastati ugrušci krvi, koji su uzrokom moždanog udara i srčanog udara.
Tlačenje

Pokazatelj trake tlaka
Pokazivačem se prati napredovanje tlaka tijekom
mjerenja.

Napuhavanje u tijeku
U tijeku je mjerenje
Napuhavanje završeno

Hrvatski 12

Pokazatelj klasifikacije WHO-a
Svih šest odjeljaka pokazivača trake odgovara klasifikaciji
krvnog tlaka WHO-a opisanoj na 14. stranici.
Primjer
Umjerena
hipertenzija

Blaga hipertenzija

Pokazatelj klasifikacije WHO-a
Teška hipertenzija
Umjerena hipertenzija
Blaga hipertenzija
Visok normalan krvni tlak
Normalan krvni tlak
Optimalan krvni tlak

Visok normalan
krvni tlak

: pokazivačem se prikazuje odjeljak,na temelju trenutačnih podataka,koji odgovara klasifikaciji WHO-a.

Što je krvni tlak?

O krvnom tlaku

Krvni tlak je pritisak krvi na stijenke arterija. Sistolički tlak nastaje kada se srce steže. Dijastolički tlak nastaje kadase srce širi. Krvni tlak mjeri se u milimetrima žive (mmHg). Prirodni krvni tlak osobe jest osnovni tlak, koji se mjeri
odmah nakon ustajanja, kada osoba još miruje i prije jela.

Što je hipertenzija i kako se kontrolira?

Ako se hipertenzija, abnormalno visok arterijski krvni tlak, ne liječi, može prouzročiti mnogo zdravstvenih problema
uključujući moždani udar i srčani udar. Hipertenziju možete kontrolirati promjenom načina života, izbjegavanjem
stresa i lijekovima pod nadzorom liječnika.
Da biste spriječili hipertenziju ili je držali pod kontrolom:
Ne pušite
Redovito vježbajte
Smanjite unos soli i masti
Idite na redovite kontroleOdržavajte pravilnu tjelesnu težinu
Hrvatski 13

Zašto mjeriti krvni tlak kod kuće?

Standarde za procjenu visokog krvnog tlaka, bez obzira na dob,
utvrdila je Svjetska zdravstvena organizacija (WHO), kako je
prikazano u tablici zdesna.

Varijacije krvnog tlaka

Krvni tlak pojedinca uvelike varira ovisno o dobu dana i godišnjem
dobu. Može varirati od 30 do 50 mmHg zbog različitih uvjeta tijekom
dana. Varijacije su još naglašenije kod pojedinaca koji imaju
hipertenziju. Normalno je da krvni tlak raste na poslu ili u igri, atijekom spavanja padne na najniže razine. Dakle, nemojte se previše
zabrinjavati rezultatima jednog mjerenja.
Mjerite u isto vrijeme svakoga dana slijedeći postupak opisan u
ovom priručniku da biste upoznali svoj normalan krvni tlak.
Redovitim očitanjima dobivate iscrpniju povijest krvnog tlaka.
Svakako zapišite datum i sat kada zabilježite krvni tlak. Posavjetujte
se s liječnikom da biste protumačili podatke o krvnom tlaku.

Hypertension 1999, Vol 17 No. 2
Hipertenzija III. stadija (teška)
Hipertenzija II. stadija (umjerena)
Hipertenzija I. stadija (blaga)
Visok normalan krvni tlak
Normalan krvni tlak
Optimalan krvni tlak

Sistolički krvni tlak
Tipična odstupanja tijekom dana
(mjereno svakih pet minuta)
Sistolički

Krvni tlak

Klasifikacija krvnog tlaka prema WHO-u

Dijastolički krvni tlak

Kada se krvni tlak mjeri u klinici ili u liječničkoj ordinaciji, osoba može osjetiti napetost i može doći do povišenih
očitanja, 25 do 30 mmHg više od vrijednosti mjerenih kod kuće. Prilikom mjerenja kod kuće smanjuju se posljedicevanjskih utjecaja na očitanja krvnog tlaka, dopunjavaju se liječnička očitanja i dobiva se točnija i potpunija povijest
krvnog tlaka.
Referentni materijal: Journal of

Dijastolički

PopodneHrvatski 14

Spavanje

UjutroVrijemeRješavanje problema
ProblemMogući razlog
Baterije su ispražnjene.

Ništa se ne pojavljuje
na zaslonu, čak i kada Terminali baterija nisu u pravilnom
je uređaj uključen.
položaju.

Ne napuhuje se
manšeta.

Uređaj ne mjeri.
Očitanja su previsoka
ili preniska.

OstaloNapomena:

Prenizak napon baterije.
(oznaka LOW BATTERY(BATERIJA PRI KRAJU)) treperi.
Ako su baterije potpuno ispražnjene, ta
se oznaka ne pojavljuje.
Manšeta nije pravilno stavljena.
Pomaknuli ste zapešće ili tijelo tijekom
mjerenja.
Položaj manšete nije dobar.

Vrijednost nije ista kao ona izmjerena u
klinici ili u liječničkoj ordinaciji.

Preporučena radnja
Zamijenite sve baterije novima.
Ponovno stavite baterije na način da se negativni
i pozitivni terminal podudaraju s onima
naznačenima na odjeljku za bateriju.

Zamijenite sve baterije novima.

Pravilno stavite manšetu.
Trebate biti vrlo mirni i tihi tijekom mjerenja.
Sjednite udobno i mirno.
Stavite ruku na stol s dlanom okrenutim prema
gore, a manšeta treba biti u razini srca.
Ako imate vrlo slabe ili nepravilne otkucaje srca,možda nastanu poteškoće za uređaj prilikomodređivanja krvnog tlaka.
Pogledajte odjeljak »Zašto mjeriti krvni tlak kod
kuće?«.
Izvadite baterije. Vratite ih unutra na pravilan
način i pokušajte ponovno s mjerenjem.

ako gore navedenim radnjama ne riješite problem, obratite se prodavaču. Ne pokušavajte sami otvoriti ili
popraviti proizvod, budući da bilo kojim pokušajem poništavate jamstvo.
Hrvatski 15Održavanje
Ne otvarajte uređaj. U njemu su osjetljivi električni dijelovi i složena zračna jedinica koje biste mogli oštetiti. Ako nemožete riješiti problem koristeći se uputama za rješavanje problema, obratite se ovlaštenom prodavaču blizu vas ili
našem odjelu korisničke službe. Korisnička služba poduzeća A&D pruža tehničke informacije, rezervne dijelove ijedinice ovlaštenim prodavačima.
Uređaj je osmišljen i proizveden da bi imao dug radni vijek. Ipak, općenito se preporučuje pregled uređaja svake
dvije godine radi jamčenja pravilnog funkcioniranja i točnosti. Obratite se ovlaštenom prodavaču blizu vas ili
poduzeću A&D radi održavanja.

Tehnički podatci
Tip
Mjerna metoda
Mjerni raspon

Mjerna točnost
Napajanje
Broj mjerenja
Opseg zapešća
Klasifikacija
Primijenjeni dio
Koristan radni vijek

UB-525
Oscilometrijsko mjerenjeTlak:
0 – 299 mmHg
Sistolički tlak:
60 – 279 mmHg
Dijastolički tlak:
40 – 200 mmHg
Puls:
40 – 180 otkucaja u minuti
Tlak:±3 mmHgPuls:±5 %2 x alkalne baterije od 1,5 V (LR03 ili AAA)
Otprilike 250 mjerenja, kada se rabe alkalne baterije AAA, uz vrijednost tlaka od
170 mmHg pri sobnoj temperaturi od 23 °C.
13,5 – 21,5 cm
Medicinska oprema s internim napajanjem (način stalnog rada)
Manšeta tip BF
Uređaj: pet godina (kada se rabi šest puta dnevno)
Hrvatski 16

Klinički test
EMD
Memorija
Radni uvjeti
Prijevozni/skladišni uvjeti
DimenzijeTežinaZaštita od prodora

U skladu s normom ISO81060-2 : 2013
U kliničkom validacijskom ispitivanju upotrijebljen je K5 na 85 ispitanika radiodređivanja dijastoličkog krvnog tlaka.
IEC 60601-1-2: 2014
Posljednjih 60 mjerenja
od +10 do +40 °C / od 15 do 85 % rel. vlaž. / od 800 do 1060 hPa
od -20 do +60 °C / od 10 do 95 % rel. vlaž. / od 700 do 1060 hPa
Pribl. 56 [Š] x 88 [V] x 21,5 [D] mm
Pribl. 90 g, ne uključujući baterije
IP20

Napomena: Specifikacije su podložne promjeni radi poboljšanja bez prethodne najave.
Klasifikacija IP-a odnosi se na stupnjeve zaštite koju pružaju kućišta u skladu s normom IEC 60529. Ovaj
je uređaj zaštićen od krutih stranih predmeta promjera 12 mm i više, primjerice prstiju. Ovaj uređaj nije
zaštićen od vode.

Hrvatski 17

MEMO

Hrvatski 18

Cuprins

Stimați clienți........................................................................................................................................................ 2
Observații preliminare .......................................................................................................................................... 2Măsuri de precauție ............................................................................................................................................. 2
Identificare piese.................................................................................................................................................. 4
Simboluri .............................................................................................................................................................. 5
Utilizarea monitorului ........................................................................................................................................... 7
1. Montarea/Înlocuirea bateriilor ..................................................................................................................... 7
2. Aplicarea manșetei ..................................................................................................................................... 8
3. Cum să faceți măsurări corecte.................................................................................................................. 8
4. Măsurare .................................................................................................................................................... 9
5. După măsurare .......................................................................................................................................... 9Măsurători .......................................................................................................................................................... 10
Reapelarea datelor din memorie ....................................................................................................................... 11Ștergerea datelor salvate în memorie ................................................................................................................ 11
Ce este indicatorul IHB/AFib? ............................................................................................................................ 12
Ce este AFib? .................................................................................................................................................... 12
Indicator bară de presiune ................................................................................................................................. 12
Indicator de clasificare OMS .............................................................................................................................. 13
Despre tensiunea arterială................................................................................................................................. 13
Ce este tensiunea arterială? ........................................................................................................................ 13
Ce este hipertensiunea și cum este controlată? .......................................................................................... 13
De ce să măsurați tensiunea arterială acasă? ............................................................................................. 14
Clasificarea tensiunii arteriale conform OMS ............................................................................................... 14
Variațiile tensiunii arteriale ........................................................................................................................... 14
Depanarea ......................................................................................................................................................... 15Întreținerea......................................................................................................................................................... 16
Date tehnice....................................................................................................................................................... 16

Română

1

Stimați clienți
Felicitări pentru achiziția monitorului de tensiune arterială A&D de ultimă generație. Proiectat pentru utilizare ușoară
și precizie, acest dispozitiv va facilita regimul zilnic de tensiune arterială.
Vă recomandăm să citiți acest manual cu atenție înainte de a utiliza dispozitivul pentru prima dată.

Observații preliminare
Acest dispozitiv este conform cu Directiva Europeană 93/42 CEE pentru produse medicale. Acest lucru este indicat. (0123: numărul de referință pentru organizația notificată implicată)de marcajul de conformitate
Dispozitivul este proiectat pentru utilizare la adulți, nu la nou-născuți sau sugari.
Mediu de utilizare. Dispozitivul poate fi utilizat de dvs. pentru îngrijire medicală la domiciliu.
Acest dispozitiv este proiectat pentru a măsura tensiunea arterială și frecvența pulsului persoanelor în scopul diagnosticării.

Măsuri de precauție

La construcția acestui dispozitiv sunt utilizate componente de precizie. Extremele de temperatură, umiditate, lumină
directă a soarelui, șoc sau praf trebuie evitate.
Curățați dispozitivul cu o cârpă uscată, moale sau o cârpă umezită cu apă și un detergent neutru. Nu utilizați
niciodată alcool, benzen, diluant sau alte substanțe chimice dure pentru a curăța dispozitivul.
Evitați să pliați strâns manșeta pentru perioade lungi, deoarece astfel de tratament poate reduce durata de viață acomponentelor.
Dispozitivul nu sunt rezistente la apă. Preveniți murdărirea dispozitivului cu ploaie, transpirație și apă.
Măsurătorile pot fi distorsionate dacă dispozitivul este utilizat aproape de televizoare, cuptoare cu microunde,telefoane mobile, dispozitive cu raze X sau alte dispozitive cu câmpuri electrice puternice.
Echipamentul, piesele și bateriile uzate nu sunt tratate ca deșeuri menajere obișnuite și trebuie eliminate în
conformitate cu reglementările locale aplicabile.
Când reutilizați dispozitivul, confirmați că este curat.
Nu modificați dispozitivul. Aceasta poate cauza accidente sau deteriora dispozitivul.
Pentru a măsura tensiunea arterială, manșeta trebuie strânsă pe încheietură suficient pentru a opri temporar fluxul
sanguin prin arteră. Aceasta poate cauza durere, amorțeală sau un semn roșu temporar pe încheietură. Această
stare apare în special atunci când măsurarea este repetată succesiv. Orice durere, amorțeală sau urme roșii vordispărea cu timpul.
Dispozitivele de comunicare wireless, cum ar fi dispozitivele de conectare în rețea la domiciliu, telefoane mobile,
Română 2

telefoane fără fir și stațiile lor de bază, stații de emisie-recepție, pot afecta acest monitor de tensiune arterială. Prinurmare, trebuie păstrată o distanță minimă de 30 cm față de astfel de dispozitive.
Măsurarea prea frecventă a tensiunii arteriale poate cauza rănirea din cauza interferenței cu fluxul sanguin.
Verificați dacă funcționarea dispozitivului nu rezultă în afectarea prelungită a circulației sângelui atunci când utilizați
dispozitivul în mod repetat.
Testarea clinică nu a fost efectuată la nou-născuți și gravide. Nu utilizați la nou-născuți sau gravide.
Dacă ați fost supusă unei mastectomii, consultați un medic înainte de a utiliza acest dispozitiv.
Nu lăsați copiii să utilizeze dispozitivul singuri și nu utilizați dispozitivul într-un loc aflat la îndemâna copiilor.
Dispozitivul poate cauza accidente sau deteriorare.
Există piese mici care pot prezenta pericol de sufocare dacă sunt înghițite din greșeală de către copii.
Nu atingeți bateriile și pacientul în același timp. Aceasta poate conduce la electrocutare.
În cazul defectării componentelor individuale, carcasa de lână manșetă se poate încălzi și poate cauza o defecțiune.
Utilizarea accesoriilor în moduri care nu sunt detaliate în acest manual poate compromite siguranța.
În caz de scurtcircuit al bateriei, aceasta se poate înfierbânta și poate cauza arsuri.
Lăsați dispozitivul să se adapteze la mediul înconjurător înainte de utilizare (aproximativ o oră).
Nu umflați fără a înfășura manșeta în jurul încheieturii.

Contraindicații

Mai jos sunt prezentate măsuri de precauție pentru utilizarea corectă a dispozitivului.
Nu aplicați dispozitivul pe o încheietură pe care este atașat alt echipament medical electric. Este posibil ca
echipamentul să nu funcționeze corect.
Persoanele care au un deficit de circulație grav în braț trebuie să consulte un medic înainte de a utiliza dispozitivul
pentru a evita problemele medicale.
Nu diagnosticați singur rezultatele măsurării și nu începeți tratamentul singur. Consultați întotdeauna medicul
pentru evaluarea rezultatelor și tratament.
Nu aplicați dispozitivul pe o încheietură cu o rană nevindecată.
Nu aplicați dispozitivul pe un braț în care se administrează o perfuzie intravenoasă sau o transfuzie de sânge.
Aceasta poate cauza rănire sau accidente.
Nu utilizați dispozitivul în spații în care sunt prezente gaze inflamabile, cum ar fi gazele de anestezie. Aceasta poate
cauza o explozie.
Nu utilizați dispozitivul în medii cu o concentrație foarte ridicată de oxigen, cum ar fi o cameră de oxigen la înaltă
presiune sau un cort cu oxigen. Aceasta poate cauza incendiu sau explozie.
Română

3

Identificare pieseAfișajButon(medie, memorie și schimbare parametri predefiniți)Buton START (pornire)
Capacul baterieiManșetăMEMORIE
Medie

Baterii alcaline (LR03 sau AAA)AfișajPresiune sistolică
Presiune diastolică

Indicator de clasificare OMSșiIndicator bară de presiune

Simbol IHB/AFib

Marcaj inimă

Frecvență puls

Indicator baterie
Română

4

Simboluri
Simboluri care sunt tipărite pe carcasa dispozitivului
Simboluri
SYS
DIA
PUL

SN

IP

Funcție / Semnificație
Pentru a porni și opri dispozitivul
Tensiune arterială sistolică în mmHg
Tensiune arterială diastolică în mmHg
Puls: număr de pulsații cardiace pe minut
Reper pentru montarea bateriei
Curent continuu
Serie
Data fabricației
Tip BF: dispozitivul și manșeta sunt proiectate pentru a asigura protecția specială împotriva șocurilor
electrice.
Dispozitiv medical conform cu directiva europeană
Simbol de protecție internațional
Etichetă Deșeuri de echipamente electrice și electronice (DEEE)
Producător
Reprezentant UE
Consultați manualul de instrucțiuni/broșura
A se feri de umiditate

Română

5

Simbolurile care apar pe afișaj
SimboluriFuncție / Semnificație / Acțiune recomandată
Apare în timp ce măsurătoarea este în curs. Luminează intermitent când este detectat pulsul.
Rămâneți cât mai calm posibil.
Simbolul IHB/AFib apare când este detectată o bătaie neregulată a inimii.
Se poate aprinde când este detectată o vibrație foarte ușoară, cum ar fi frisoane sau tremurat.

Simbolurile care apar pe afișaj (continuare)
SimboluriFuncție / Semnificație
Măsurările anterioare salvate în MEMORIE
Date medii
BATERIE PLINĂ
Indicatorul de alimentare a bateriei în timpul măsurăriiBATERIE SLABĂ
Tensiunea bateriei este joasă atunci când clipește

sau

Tensiune arterială instabilă din cauza mișcării în timpulmăsurării

Acțiune recomandată

Înlocuiți toate bateriile cu unele noi
atunci când indicatorul luminează
intermitent.
Efectuați altă măsurare.
Rămâneți nemișcat în timpulmăsurării.

Diferența dintre valorile sistolice și diastolice este mai mică de
10 mmHg.
Valoarea presiunii nu a crescut în timpul umflării.
Manșeta nu este aplicată corect.
EROARE AFIȘAJ PUL
Pulsul nu este detectat corect.

Aplicați manșeta corect și efectuați
altă măsurare.

Eroare internă de monitor de tensiune arterială

Scoateți bateriile și apăsați butonul
START apoi instalați bateriile din nou.
Dacă eroarea mai apare, contactați
dealerul.

Română

6

Utilizarea monitorului

1. Montarea/Înlocuirea bateriilor
1. Scoateți capacul bateriilor.

2. Scoateți bateriile uzate și introduceți bateriile noi în
compartimentul pentru baterii după cum este indicat, avândgrijă ca polaritățile (+ și -) să fie corecte. Utilizați doar bateriiLR03 sau AAA.
3. Montați capacul bateriei.

ATENȚIONĂRI

Capacul bateriei

Baterii uzate

Baterii noi
Introduceți bateriile după cum este indicat în compartimentul
pentru baterii. Dacă sunt montate incorect, dispozitivul nu vafuncționa.
Când simbolul(marcajul BATERIE SLABĂ) luminează
intermitent pe afișaj, înlocuiți toate bateriile cu altele noi. Nu
amestecați bateriile vechi cu cele noi. Aceasta poate reduce
durata de viață a bateriei sau poate cauza defectarea
dispozitivului.
(marcajul BATERIE SLABĂ) nu apare când bateriile
sunt epuizate.
Durata de viață a bateriei variază în funcție de temperatura ambiantă și poate fi mai scurtă la temperaturi joase. În
general, două baterii LR03/AAA noi vor dura aproximativ patru luni atunci când sunt utilizate pentru două măsurători pe zi.
Utilizați doar baterii specificate. Bateriile furnizate împreună cu dispozitivul sunt pentru testarea performanței
monitorului și pot avea o durată de viață limitată.
Scoateți bateriile dacă dispozitivul nu va fi utilizat o perioadă îndelungată de timp. Bateriile pot prezenta scurgeri și
pot cauza defectarea.
Română 7

2. Aplicarea manșetei
1. Înfășurați manșeta în jurul încheieturii, la aproximativ 1 cm deasupra mâinii, ca înfigura din dreapta.

Aproximativ 1 cm

2. Aplicați manșeta strâns utilizând banda Velcro.
Notă: Pentru măsurători adecvate, aplicați manșeta strâns și măsurați pe oîncheietură goală.

3. Cum să faceți măsurări corecte

Pentru cea mai corectă măsurare a tensiunii arteriale:
Nu vă mișcați și păstrați-vă calmul în timpul măsurării.
Așezați-vă confortabil în șezut. Puneți cotul pe masă cu palma în sus și manșeta la același nivel cu inima.
Relaxați-vă aproximativ cinci până la zece minute înainte de măsurare. Dacă sunteți enervat(ă) sau deprimat(ă) din
cauza stresului emoțional, măsurarea va reflecta acest stres sub forma unei citiri a tensiunii arteriale mai mari (saumai mici) decât cea normală și valoarea pulsului va fi de obicei mai rapidă decât cea normală.
Încercați să măsurați tensiunea arterială aproximativ în același timp în fiecare zi.
Tensiunea arterială a unei persoane variază constant. Ce faceți, ce ați mâncat și ce beți poate avea un efect foarte
puternic și rapid asupra tensiunii dvs. arteriale.
Nu măsurați imediat după efort fizic sau o baie. Înainte de măsurare, odihniți-vă timp de douăzeci până la treizeci deminute.
Nu vă încrucișați picioarele. Țineți picioarele pe sol și îndreptați spatele.
Acest dispozitiv își bazează măsurătorile pe bătăile inimii. Dacă bătăile inimii sunt foarte slabe sau neregulate,dispozitivul poate întâmpina dificultăți în determinarea tensiunii arteriale.
Dacă dispozitivul detectează o stare anormală, acesta va opri măsurarea și va afișa un simbol de eroare. Consultați
pagina 6 pentru descrierea simbolurilor.
Acest monitor de tensiune arterială este conceput pentru a fi utilizat de adulți. Consultați medicul înainte de a utiliza
acest dispozitiv la un copil. Un copil nu trebuie să utilizeze acest dispozitiv nesupravegheat.
Performanța monitorului automat de tensiune arterială poate fi afectată de temperatura sau umiditatea excesivă,sau de altitudine.
Română 8

4. MăsurareÎn timpul măsurării, este normal să simțiți că manșeta este foarte strânsă.

5. După măsurareÎn timpul afișării măsurătorilor, dacă apăsați butonul START pentru a opri dispozitivul, măsurătorile noi suntmemorate.
În timpul afișării măsurătorilor, dacă apăsați butonul
pentru a opri dispozitivul, măsurătorile noi sunt memorate.
Scoateți manșeta și înregistrați datele.
Notă:dispozitivul este dotat cu funcție de oprire automată a alimentării, care memorează datele curente șioprește automat dispozitivul după un minut de la măsurare. Așteptați cel puțin trei minute între măsurători
la aceeași persoană.

Română

9Măsurători
Notă: după ce este utilizat, UB-525 va asigura umflarea adecvată utilizatorului.
1. Înfășura manșeta în jurul încheieturii. Stați confortabil, cu manșeta la același
nivel cu inima dvs. și relaxați-vă.
2. Apăsați butonul START . Apar toate segmentele afișajului.
3. Apare zero (0) luminând intermitent pentru scurt timp. Apoi, afișajul se schimbă,în timp ce începe măsurarea. Manșetele încep să se umfle. Este normal săsimțiți că manșeta este foarte strânsă. Măsurarea începe automat când începe
umflarea și(marcajul inimă) luminează intermitent.
Notă:dacă doriți să opriți umflarea în orice moment, apăsați butonul
START din nou.

4. Când măsurarea este finalizată, dispozitivul afișează măsurători (valorile
presiunii sistolice și diastolice, frecvența pulsului, clasificarea OMS simbolul
IHB). Manșeta se dezumflă complet automat, evacuând aerul rămas.
Notă: dacă nu doriți să memorați măsurători noi, apăsați butonul
sunt afișate măsurătorile.

în timp ce

5. După măsurare, apăsați butonul START pentru a opri dispozitivul.
Scoateți manșonul.
Notă: dispozitivul este prevăzut cu funcție de oprire automată a alimentării.
Așteptați cel puțin trei minute între măsurători la aceeași persoană.

Română

10

Toate segmenteleAfișaj zero
Umflarea a început
Umflare și măsurareîn curs
Presiune sistolică
Presiune diastolică
Clasificare OMS
Simbol IHB
Frecvență puls

Reapelarea datelor din memorie
Notă: acest dispozitiv salvează ultimele 60 de măsurători în memorie.
Mod memoriepentru a opri dispozitivul.
1. Apăsați butonul
Media tuturor măsurătorilor și numărul de date sunt afișate. Dacă nu
există date, se afișează „0”. Apăsați butonul
sau START pentru aMedia tuturor datelor
opri dispozitivul.
Numărul tuturor datelor
2. Utilizați următoarele butoane pentru a afișa date (date de măsurare șinumăr).
Dispozitivul afișează media tuturor măsurătorilor și numărul de datesunt afișate.
Ex.: No.30
Cele mai recente dateApăsând butonul
de fiecare dată, dispozitivul este afișat dupăcum urmează:Date medii ale tuturor măsurătorilor intermediare.
Ultimele date
Date (date de măsurare și număr).
(Cele
mai vechi)
Dispozitivul afișează în ordine de la cele mai recente date.
În exemplu:
No.30 și date
No.29 și date
No.01 și date.
după ce sunt afișate cele mai vechi date,
3. Dacă apăsați butonul
dispozitivul trece la pasul 1, sunt afișate media tuturor măsurătorilor șinumărul datelor.
4. Apăsați butonul START pentru a opri dispozitivul.
După un minut de nefuncționare, dispozitivul se va opri automat.

Ștergerea datelor salvate în memorie

1. Apăsați lung butonulpână când dispozitivul se oprește automat.

, șterge datele memorate în timp ce marcajul
Dispozitivul afișează marcajul
luminează intermitent și se oprește automat.
Română 11

Oprire
Buton

Buton

Buton

Buton

OprireCe este indicatorul IHB/AFib?
Când monitorul detectează un ritm neregulat în timpul măsurării, indicatorul IHB/AFib va apărea pe afișaj cu
valorile măsurării.
Notă: vă recomandăm să contactați medicul dacă vedeți acest indicator IHB/AFibfrecvent.

Ce este AFib?

Inima se contractă ca urmare a semnalelor electrice care apar în inimă și trimite sânge în corp. Fibrilația atrială
(AFib) are loc atunci când semnalul electric din atriu devine confuz și duce la perturbări în intervalul impulsului. AFibpoate cauza stagnarea sângelui în inimă, ceea ce poate crea ușor cheaguri de sânge, o cauză a accidentului
vascular cerebral și a infarctului miocardic.
Presurizare

Indicator bară de presiune

Indicatorul monitorizează evoluția presiunii în timpulmăsurării.
Umflare în cursMăsurare în curs
Umflare completă

Română

12

Indicator de clasificare OMS
Fiecare șase segmente ale indicatorului bară corespundclasificării OMS a tensiunii arteriale descrise la pagina 14.
Exemplu
Hipertensiune
moderată

Hipertensiuneușoară

Ridicată normală

Indicator de clasificare OMS
Hipertensiune severă
Hipertensiune moderată
Hipertensiune ușoară
Ridicată normală
Normală
Optimă
: Indicatorul afișează un segment,pe baza datelor curente,corespunzător clasificării OMS.

Ce este tensiunea arterială?

Despre tensiunea arterială

Tensiunea arterială este forța exercitată de sânge asupra pereților arterelor. Presiunea sistolică apare atunci cândinima se contractă. Presiunea diastolică apare atunci când inima se dilată. Tensiunea arterială este măsurată în
milimetri coloană de mercur (mmHg). Tensiunea arterială naturală este reprezentată de presiunea fundamentală,care este măsurată înainte de orice activitate, dimineața la trezirea din pat și înainte de a mânca.

Ce este hipertensiunea și cum este controlată?

Hipertensiunea este o tensiune arterială anormal de ridicată. Nesupravegheată, aceasta poate provoca multe
probleme de sănătate, inclusiv accident vascular cerebral și infarct miocardic. Hipertensiunea poate fi controlată
prin schimbarea stilului de viață, evitarea stresului și cu medicamente luate sub supraveghere medicală.
Pentru a preveni hipertensiunea sau a o menține sub control:
Nu fumați
Faceți regulat efort fizic
Reduceți consumul de sare și grăsimi
Mergeți periodic la consultații medicaleMențineți o greutate adecvată
Română 13

De ce să măsurați tensiunea arterială acasă?

Standardele de evaluare a tensiunii arteriale mari, indiferent desvârstă, au fost stabilite de Organizația Mondială a Sănătății
(OMS), după cum este indicat în graficul din dreapta.

Variațiile tensiunii arteriale

Tensiunea arterială a unei persoane variază foarte mult în funcție demomentul zilei și de anotimp. Aceasta poate varia între 30 și 50
mmHg ca urmare a diferitelor condiții din timpul zilei. La persoanele
cu hipertensiune, variațiile sunt și mai pronunțate. În mod normal,
tensiunea arterială crește în timpul lucrului sau jocului și ajunge lacele mai scăzute niveluri în timpul somnului. Prin urmare, nu văîngrijorați excesiv privind rezultatele unei măsurări.
Efectuați măsurători la aceeași oră în fiecare zi, utilizând procedura
descrisă în acest manual pentru a afla tensiunea dvs. arterială
normală. Măsurătorile regulate oferă un istoric mai complex altensiunii arteriale. Asigurați-vă că notați data și ora atunci cândînregistrați tensiunea arterială. Consultați medicul pentru a interpreta
datele tensiunii arteriale.

Română

14(Jurnalul hipertensiunii) 1999, Vol. 17 Nr. 2
Hipertensiune de gradul 3 (gravă)Hipertensiune de gradul 2 (moderată)Hipertensiune de gradul 1 (ușoară)Ridicată-normală
Normală
Optimă

Tensiune arterială sistolică
Fluctuația tipică dintr-o zi(Măsurată la fiecare cinci minute)Tensiune arterială

Clasificarea tensiunii arteriale conform OMS

Tensiune arterială diastolică

Măsurarea tensiunii arteriale la clinică sau cabinetul medical poate cauza anxietate și poate produce o citire ridicată,cu 25 până la 30 mmHg mai mare decât cea măsurată acasă. Măsurarea acasă reduce efectele influențelor
exterioare asupra citirilor tensiunii arteriale, completează măsurătorile efectuate de medic și oferă un istoric mai
precis și complet al tensiunii arteriale.
Material de referință: Journal of Hypertension

Sistolică

Somn

Diastolică
PM

AM

Oră

Depanarea
Problemă

Motiv posibil
Bateriile sunt descărcate.

Acțiune recomandată
Înlocuiți toate bateriile cu altele noi.
Reinstalați bateriile cu bornele pozitive șinegative corespunzătoare celor indicate pecompartimentul pentru baterii.

Nu apare nimic peafișaj, chiar și cândBornele bateriilor nu sunt în poziția
dispozitivul este pornit. corectă.

Manșeta nu se umflă.

Tensiunea bateriei este prea mică.
(marcajul BATERIE SLABĂ)clipește. Dacă bateriile sunt descărcatecomplet, marcajul nu apare.
Manșeta nu este aplicată corect.
Ați mișcat încheietura sau corpul în
timpul măsurării.

Înlocuiți toate bateriile cu altele noi.

Aplicați manșeta corect.
Asigurați-vă că rămâneți perfect nemișcat(ă) șiliniștit(ă) în timpul măsurării.
Dispozitivul nuAșezați-vă confortabil și nu vă mișcați.
măsoară.
Puneți brațul pe masă cu palma în sus șiMăsurătorile sunt prea Poziția manșetei nu este corectă.
manșeta la același nivel cu inima.
mari sau prea mici.
Dacă bătăile inimii sunt foarte slabe sau
neregulate, dispozitivul poate avea dificultate în
determinarea tensiunii arteriale.
Valoarea este diferită de cea măsurată la Consultați secțiunea „De ce să măsurați
clinică sau cabinetul medical.
tensiunea arterială acasă?”.
Alta
Scoateți bateriile. Așezați-le corect înapoi șiîncercați măsurarea din nou.
Notă:dacă acțiunile descrise mai sus nu rezolotvă problema, contactați dealerul. Nu încercați să deschideți sau să
reparați acest produs, deoarece orice încercare a face acest lucru va anula garanția dvs.
Română

15Întreținerea
Nu deschideți dispozitivul. Acesta utilizează componente electrice sensibile și o unitate de aer complexă, care pot fi
deteriorate. Dacă nu puteți remedia problema utilizând instrucțiunile de depanare, contactați dealerul autorizat dinzona dvs. sau departamentul nostru de asistență clienți. Serviciul clienți A&D va furniza informații tehnice, piese deschimb și unități dealerilor autorizați.
Dispozitivul a fost proiectat și fabricat pentru a avea o durată de viață îndelungată. În general, este recomandat săverificați dispozitivul o dată la 2 ani pentru a asigura funcționarea și precizia corecte. Contactați dealerul autorizat
din zona dvs. sau A&D pentru întreținere.

Date tehnice
Tip
Metodă de măsurare
Interval de măsurare

Precizie de măsurare
Alimentare electrică
Număr de măsurări
Circumferința încheieturii
ClasificarePiesă aplicată
Durată de viață utilă

UB-525Măsurare oscilometrică
Presiune:0 – 299 mmHg
Presiune sistolică:60 – 279 mmHg
Presiune diastolică:40 – 200 mmHg
Puls:
40 – 180 bătăi/minut
Presiune: ±3 mmHg
Puls:
±5 %
2 baterii alcaline de 1,5 V (LR03 sau AAA)
Aprox. 250 de măsurători, când sunt utilizate baterii alcaline AAA, cu valoarea
presiunii de 170 mmHg la temperatura camerei 23 °C.
13,5 - 21,5 cm
Echipament ME alimentat intern (Mod de funcționare continuă)Manșetă tip BF
Dispozitiv: 5 ani (când este utilizat de șase ori pe zi)
Română

16

Conform ISO81060-2：2013
Test clinic
În studiul de validare clinică, K5 a fost utilizat la 85 de subiecți pentru determinarea
tensiunii arteriale diastolice.
EMD
IEC 60601-1-2: 2014
Memorie
Ultimele 60 de măsurători
Condiții de funcționare+10 până la +40 °C / 15 până la 85 % RH / 800 până la 1.060 hPa
Condiții de transport / depozitare -20 până la +60 °C / 10 până la 95 % RH / 700 până la 1.060 hPa
Dimensiuni
Aprox. 56 [l] x 88 [Î] x 21,5 [A] mm
Greutate
Aprox. 90 g, fără baterii
Protecție împotriva infiltrației
IP20
Notă:

specificațiile pot fi modificate pentru îmbunătățire fără notificare prealabilă.
Clasificarea IP înseamnă gradele de protecție asigurate de carcase în conformitate cu IEC 60529. Acestdispozitiv este protejat împotriva corpurilor străine solide cu diametrul mai mare sau egal cu 12 mm, cum ar fi
un deget. Acest dispozitiv nu este protejat împotriva apei.

Română

17

MEMO

Română

18

Sisukord

Head kliendid ....................................................................................................................................................... 2
Eelmärkused ........................................................................................................................................................ 2
Ettevaatusabinõud ............................................................................................................................................... 2
Seadme osad....................................................................................................................................................... 4
Sümbolid .............................................................................................................................................................. 5Vererõhumõõtja kasutamine ................................................................................................................................ 7
1. Patareide paigaldamine/vahetamine .......................................................................................................... 7
2. Manseti paigaldamine ................................................................................................................................ 8
3. Täpsete mõõtmistulemuste saavutamine ................................................................................................... 8
4. Mõõtmine ................................................................................................................................................... 9
5. Pärast mõõtmist ......................................................................................................................................... 9Mõõtmised ......................................................................................................................................................... 10
Andmete vaatamine mälust ............................................................................................................................... 11
Mällu salvestatud andmete kustutamine ............................................................................................................ 11
Mis on IHB/AFib-näidik? .................................................................................................................................... 12
Mis on AFib? ...................................................................................................................................................... 12Rõhu ribanäidik .................................................................................................................................................. 12
WHO klassifikatsiooni näidik .............................................................................................................................. 13
Vererõhk ............................................................................................................................................................ 13
Mis on vererõhk? .......................................................................................................................................... 13
Mis on hüpertoonia ja kuidas seda kontrollitakse? ....................................................................................... 13
Miks mõõta vererõhku kodus? ..................................................................................................................... 14
WHO vererõhu klassifikatsioon .................................................................................................................... 14
Vererõhu muutumine .................................................................................................................................... 14
Rikkeotsing ........................................................................................................................................................ 15
Hooldus.............................................................................................................................................................. 16
Tehnilised andmed ............................................................................................................................................ 16

Eesti

1

Head kliendidÕnnitleme teid kõrgtehnoloogilise A&D vererõhumõõtja ostu puhul. Seade on kujundatud, pidades silmas
kasutusmugavust ja täpsust, ning see aitab teil iga päev oma vererõhku kontrolli all hoida.
Enne kui kasutate seadet esimest korda, soovitame teil hoolikalt läbi lugeda selle juhendi.

Eelmärkused

Seade vastab Euroopa meditsiiniseadmete direktiivi 93/42 EMÜ nõuetele. Sellele viitab vastavusmärk(0123: teavitatud asutuse viitenumber)
Seade on mõeldud kasutamiseks täiskasvanutele, mitte vastsündinutele ega imikutele.
Kasutuskeskkond. Seade on mõeldud iseseisvalt kasutamiseks koduses ravikeskkonnas.
Seade on mõeldud inimeste vererõhu ja pulsisageduse mõõtmiseks diagnoosimise eesmärgil.

Ettevaatusabinõud

.

Seadmes on kasutusel täppiskomponendid. Vältida tuleb äärmuslikke temperatuure, õhuniiskust, otsest
päikesevalgust, põrutamist ja tolmu.
Puhastage seadet kuiva pehme lapiga või lapiga, mida on niisutatud vee ja neutraalse pesuainega. Ärge kasutage
seadme puhastamiseks kunagi alkoholi, benseeni, vedeldit ega muid tugevaid kemikaale.
Ärge voltige mansetti pikaks ajaks liiga tugevalt kokku, muidu võib komponentide eluiga lüheneda.
Seade pole veekindel. Vältige vihma, higi ja vee sattumist seadmesse.
Mõõtmistulemused võivad olla moonutatud, kui seadme läheduses on kasutamise ajal televiisor, mikrolaineahi,
mobiiltelefon, läbivalgustusaparaat või mõni muu tugeva elektriväljaga seade.
Kasutatud seadet, komponente ega patareisid ei tohi käidelda koos tavaliste majapidamisjäätmetega, nendest
tuleb vabaneda kohalike nõuete kohaselt.
Kui kasutate seadet korduvalt, kontrollige, kas seade on puhas.
Ärge modifitseerige seadet. See võib põhjustada õnnetuse või kahjustada seadet.
Vererõhu mõõtmiseks peab mansett rannet pigistama piisavalt tugevalt, et ajutiselt arterit läbiv verevool peatada.
See võib tekitada valu, tuimust ja jätta randmele ajutiselt punase jälje. See tingimus esineb kõige sagedamini siis,
kui mõõtmist korratakse mitu korda järjest. Valu, tuimus ja punased jäljed kaovad aja jooksul.
Vererõhumõõtjat võivad mõjutada raadiosideseadmed, nagu koduvõrguseadmed, mobiiltelefonid, traadita telefonid
Eesti 2

ja nende tugijaamad ning käsisaatjad. Seetõttu tuleb seda sellistest seadmetest hoida vähemalt 30 cm kaugusel.
Liiga sage vererõhu mõõtmine võib verevoolu häirimise tõttu olla kahjulik. Jälgige, et seadme korduva kasutusega
ei kaasneks pikaajalist vereringe häirimist.
Seadet pole kliiniliselt testitud vastsündinud imikutel ega rasedatel naistel. Ärge kasutage seadet vastsündinud
imikutel ega rasedatel naistel.
Kui teile on tehtud mastektoomia, pidage enne seadme kasutamist nõu arstiga.
Ärge laske lastel seadet iseseisvalt kasutada ja ärge kasutage seadet kohas, kus see on imikutele kättesaadav.
See võib põhjustada õnnetuse või kahjustuse.
Seadmes on väikseid osi, mille allaneelamise korral võib imik lämbuda.
Ärge puudutage samaaegselt patareisid ja patsienti. See võib põhjustada elektrilöögi.
Üksikute komponentide rikke korral võib manseti lähedal olev sulgur kuumaks muutuda ja põhjustada
talitlushäireid.
Lisatarvikute, mida selles juhendis pole kirjeldatud, kasutamine võib seada ohtu ohutuse.
Patarei lühise korral võib see muutuda kuumaks ja põhjustada põletusi.
Laske seadmel ümbritseva keskkonnaga kohaneda, enne kui seda kasutate (ligikaudu üks tund).
Ärge täitke õhuga, kui mansett pole mähitud ümber randme.

Vastunäidustused

Allpool on toodud ettevaatusabinõud seadme õigesti kasutamiseks.
Ärge kinnitage mansetti randmele, millele on juba kinnitatud mõni muu meditsiiniline elektriseade. Seade ei pruugiõigesti toimida.
Inimesed, kelle käsivarre verevarustus on raskelt häiritud, peavad enne seadme kasutamist konsulteerima arstiga,
et vältida meditsiiniliste probleemide esinemist.
Ärge diagnoosige ise mõõtmistulemusi ega alustage iseseisvalt ravi. Konsulteerige alati arstiga, et hinnatamõõtmistulemusi ja määrata ravi.
Ärge kinnitage seadet randmele, millel on paranemata haav.
Ärge kinnitage seadet käsivarrele, millele on kinnitatud veenisisene tilguti või mille kaudu tehakse vereülekannet.
See võib põhjustada vigastuse või õnnetuse.
Ärge kasutage seadet kohas, kus on tuleohtlikke gaase (nt narkoosigaas). See võib põhjustada plahvatuse.
Ärge kasutage seadet kõrge hapnikusisaldusega keskkonnas (nt kõrgsurvestatud hapnikukambris,
hapnikuseadmes). See võib põhjustada tulekahju või plahvatuse.
Eesti

3

Seadme osad
EkraanNupp (keskmine, mälu ja eelseadistatud parameetrite muutmine)

START Nupp

Patareipesa kaas

Mansett

MÄLUKeskmine

Leelispatareid (LR03 või AAA)

EkraanSüstoolne rõhkDiastoolne rõhkWHO klassifikatsiooni näidik
jarõhu ribanäidik

IHB/AFibi sümbol

Südamemärk

Pulsisagedus

Patareinäidik
Eesti

4

Sümbolid

Sümbolid, mis on trükitud seadme korpusele
Sümbolid

Funktsioon/tähendus

Puhkerežiim ja seadme sisselülitamine
SYS

Süstoolne vererõhk (mmHg)

DIA

Diastoolne vererõhk (mmHg)

PUL

Pulss minutis
Patarei paigaldusjuhend
Alalisvool

SN

Seerianumber
Tootmiskuupäev
Tüüp BF: seade ja mansett on kujundatud selliselt, et need pakuksid kaitset elektrilöökide eestEÜ direktiivile vastava meditsiiniseadme silt

IP

Rahvusvahelise kaitseastme sümbol
WEEE-silt
TootjaEL-i esindaja
Vt kasutusjuhendit/voldikut
Hoidke kuivana

Eesti

5

Sümbolid, mida kuvatakse ekraanil
Sümbolid

Funktsioon/tähendus/soovituslik tegevus
Kuvatakse, kui mõõtmine on pooleli. Vilgub, kui tuvastatakse pulss.
Püsige võimalikult liikumatult.
IHB/AFibi sümbol kuvatakse südame rütmihäire tuvastamise korral.
See võib süttida, kui tuvastatakse väga õrn vibreerimine (nt värisemine või rappumine).

Sümbolid, mida kuvatakse ekraanil (jätkub)
Sümbolid

Funktsioon/tähendusMÄLLU salvestatud eelnevad mõõtmised
Keskmised andmedTÄIS PATAREI
Patarei laetusnäidik mõõtmise ajalTÜHI PATAREI
Patarei on tühi, kui see vilgub

Soovituslik tegevus

Ebastabiilne vererõhk tulenevalt liikumisest mõõtmise ajalvõi

Asendage kõik patareid uutega, kui
see näidik vilgub.
Mõõtke uuesti.
Püsige mõõtmise ajal liikumatult.

Süstoolse ja diastoolse vererõhu väärtused erinevad üksteisest
kuni 10 mmHg
Rõhk ei suurenenud pumpamise ajal
Kinnitage mansett korralikult jamõõtke uuesti.
Mansett ei ole korralikult kinnitatud
PULSINÄIDU VIGA
Pulssi ei tuvastatud õigesti
Eemaldage patareid, vajutage nuppu
START ja seejärel paigaldage patareid
Vererõhumõõtja sisemine viga
tagasi. Kui viga kuvatakse uuesti,
pöörduge edasimüüja poole.
Eesti

6

Vererõhumõõtja kasutamine

1. Patareide paigaldamine/vahetamine
1. Eemaldage patareipesa kaas.

2. Eemaldage kasutatud patareid ja paigaldage uued patareidnäidatud viisil patareipessa, pidades silmas, et nendepolaarsus oleks õige (+ ja –). Kasutage ainult patareisid LR03
või AAA.
3. Paigaldage patareipesa kaas.

Patareipesa kaas

Kasutatud patareid

ETTEVAATUST

Uued patareid

Paigaldage patareid näidatud viisil patareipessa.
Kui need on valesti paigaldatud, ei hakka seade tööle.
Kui märk
(TÜHI PATAREI) ekraanil vilgub, asendage
kõik patareid uutega. Ärge segage omavahel vanu ja uusi
patareisid. See võib patareide eluiga lühendada võipõhjustada seadme talitlushäire.
Märki(TÜHI PATAREI) ei kuvata, kui patareid ontäiesti tühjad.
Patarei eluiga erineb olenevalt keskkonnatemperatuurist ja võib külmemas temperatuuris olla lühem. Tavaliselt
peavad kaks uut patareid LR03/AAA vastu ligikaudu neli kuud, kui neid kasutatakse iga päev mõõtmiseks kaks
korda.
Kasutage ainult lubatud patareisid. Seadmega kaasnevad patareid on mõeldud vererõhumõõtja proovimiseks ja
nende eluiga võib olla piiratud.
Eemaldage patareid, kui seadet ei kasutata pikema aja jooksul. Patareid võivad lekkida ja põhjustada talitlushäire.
Eesti

7

2. Manseti paigaldamine
1. Keerake mansett ümber randme ligikaudu 1 cm käelabast kõrgemal, nagu on
näidatud paremal oleval joonisel.

Ligikaudu 1 cm

2. Paigaldage mansett takjariba abil tihedalt.
Märkus. Täpsete mõõtmistulemuste saamiseks kinnitage mansett tihedalt ja mõõtke
paljalt randmelt.

3. Täpsete mõõtmistulemuste saavutamine

Vererõhu võimalikult täpseks mõõtmiseks toimige alljärgnevalt:
Olge mõõtmise ajal liikumatu ja vaikne.
Istuge mugavas asendis. Asetage pihk ülespoole suunatult oma küünarnukk lauale nii, et mansett on südamega
samal kõrgusel.
Lõõgastuge enne mõõtmist ligikaudu 5–10 minutit. Kui olete erutunud või emotsionaalselt kurnatud, väljendub see
stress tavalisest kõrgemas (või madalamas) vererõhunäidus ning pulss on üldiselt tavapärasest kiirem.
Püüdke iga päev mõõta vererõhku samal ajal.
Inimese vererõhk muutub pidevalt – see, mida ta teeb, mida ta on söönud ja joonud, võib väga tugevalt ja kiirestimõjutada vererõhku.
Ärge mõõtke vahetult pärast füüsilist treeningut ega vanniskäiku. Puhake enne mõõtmist 20–30 minutit.
Ärge ristake jalgu. Hoidke tallad vastu põrandat ja selg sirge.
See seade tuletab mõõtmistulemused südamelöögist lähtuvalt. Kui teie südamelöögid on väga nõrgad võiebaregulaarsed, võib seadmel olla raske vererõhku määrata.
Kui seade tuvastab tavatu tingimuse, katkestab see mõõtmise ja kuvab veasümboli. Sümbolite kirjeldusi vt lk 6.
See vererõhumõõtja on mõeldud kasutamiseks täiskasvanutele. Enne seadme kasutamist lapsel pidage nõuarstiga. Laps ei tohi seadet kasutada järelevalveta.
Automaatse vererõhumõõtja tööd võib mõjutada äärmuslik temperatuur, õhuniiskus või kõrgus merepinnast.
Eesti

8

4. MõõtmineMõõtmise ajal on normaalne, et mansett tundub väga pingul olevat.

5. Pärast mõõtmist

Kui kuvatud on näidud ja te vajutate seadme väljalülitamiseks nuppu
Kui kuvatud on näidud ja te vajutate seadme väljalülitamiseks nuppu
Eemaldage mansett ja salvestage andmed.

START , salvestatakse uued näidud mällu.

, ei salvestata uusi näite.

Märkused. Seadmel on automaatne väljalülitusfunktsioon, mis salvestab praegused andmed mälus ja lülitab
seadme automaatselt välja ligikaudu üks minut pärast mõõtmist. Jätke sama inimese mõõtmiste vahele
vähemalt kolm minutit.

Eesti

9

Mõõtmised
Märkus. Pärast esimest kasutamist täidab UB-525 seadme kasutaja jaoks sobiva
rõhuga.
1. Mähkige mansett ümber randme. Istuge mugavalt nii, et mansett on teiesüdamega samal kõrgusel, ja lõõgastuge.
2. Vajutage nuppu

START . Kuvatakse kõik ekraani segmendid.

3. Hetkeks kuvatakse vilkuv null (0). Kui alustatakse mõõtmist, siis kuva muutub.
Mansett täidetakse õhuga. On normaalne, et mansett tundub väga pingul olevat.
Mõõtmine algab koos õhuga täitmisega automaatselt ja südamemärk
hakkab vilkuma.
Märkus. Õhuga täitmise katkestamiseks ükskõik millisel hetkel vajutage uuesti
nuppu START .
4. Kui mõõtmine on lõpetatud, kuvatakse seadme ekraanil näidud (süstoolse ja
diastoolse vererõhu näidud, pulsisagedus, WHO klassifikatsioon, IHB sümbol).
Mansetist lastakse automaatselt välja kogu sellesse jäänud õhk.

Kõik segmendid
Nulli kuvaÕhuga täitmist on
alustatudÕhuga täitmine jamõõtmine on pooleli

Süstoolne rõhkDiastoolne rõhkMärkus. Kui te uusi näite mällu salvestada ei soovi, vajutage näitude kuvamise
WHO klassifikatsioon
ajal nuppu .
IHB sümbol
5. Vajutage uuesti nuppu START , et seade välja lülitada. Eemaldage
Pulsisagedus
mansett.
Märkused. Seadmel on automaatne väljalülitusfunktsioon.
Jätke sama inimese mõõtmiste vahele vähemalt kolm minutit.

Eesti

10

Andmete vaatamine mälust
Väljalülitamine
Märkus.
Seade salvestab viimased 60 mõõtmist mällu.
Mälurežiim
1. Vajutage seadme väljalülitamisel nuppu .
Nupp
Kuvatakse kõigi mõõtmiste keskmine ja andmete arv. Andmete
puudumise korral kuvatakse „0”. Vajutage nuppu
või START , etKõigi andmete keskmine
seade välja lülitada.
Kõigi andmete arv
2. Andmete kuvamiseks kasutage järgmisi nuppe (arvandmed ja
Nuppmõõtmisandmed).
Seade kuvab kõigi mõõtmiste keskmise ja andmete arvu.
Nt: nr 30
Nupu
vajutamise korral, kuvab seade järgnevalt järgmist.
Kõige uuemad andmedKõigi vahemikus tehtud mõõtmiste keskmised andmed.
Andmed (arvandmed ja mõõtmisandmed).
Nupp
Seade alustab kuvamist kõige uuematest andmetest.
Viimased andmed
Näites: nr 30 ja andmed
nr 29 ja andmed
nr 01 ja andmed.
(vanimad)
3. Kui vajutate pärast vanimate andmete kuvamist nuppu , jätkab seadeNupp
sammust 1, kuvades kõigi mõõtmiste keskmise ja andmete arvu.

4. Vajutage nuppu START , et seade välja lülitada.
Kui ühe minuti jooksul ei tehta ühtegi toimingut, lülitub seade automaatselt
välja.

Mällu salvestatud andmete kustutamine
1. Vajutage ja hoidke all nuppu, kuni seade automaatselt välja lülitatakse.

, kustutab mällu salvestatud andmed, vilgutades märki
Seade kuvab märgi, ja lülitub automaatselt välja.
Eesti 11

Väljalülitamine

Nupp

Mis on IHB/AFib-näidik?
Kui vererõhumõõtja tuvastab mõõtmistemõõtmistulemustega IHB/AFib-näidik.

ajalebaregulaarse

Märkus. Kui näete ekraanil sageli IHB/AFib-näidikut

südamerütmi,kuvatakse

ekraanilkoos, soovitame pöörduda arsti poole.

Mis on AFib?

Südamelihas tõmbub südamest lähtuva elektriimpulsi mõjul kokku ja pumpab kogu kehasse verd. Atriaalne
fibrillatsioon (AFib) leiab aset, kui südamekojast lähtuv elektriimpulss tekitab segadust ja muudab pulsisageduseebaregulaarseks. AFib võib põhjustada verevoolu seiskumist südames, millega võib omakorda kaasneda trombiteke ja rabandus või südameinfarkt.
SurvestamineRõhu ribanäidik

Näidik jälgib rõhu muutumist mõõtmise ajal.
Õhuga täitmine on pooleliMõõtmine on pooleliÕhuga täitmine on lõpetatud

Eesti

12

WHO klassifikatsiooni näidik
Ribanäidiku kuus segmenti vastavad WHO vererõhu
klassifikatsioonile, mida on lähemalt kirjeldatud lk 14.
NäideMõõdukashüpertooniatõbi

Kerge
hüpertooniatõbi

Kõrge normaalne

WHO klassifikatsiooni näidik
Raske hüpertooniatõbi
Mõõdukas hüpertooniatõbi
Kerge hüpertooniatõbi
Kõrge normaalne
Normaalne
Optimaalne: näidik kuvab segmenteolemasolevate andmete kohaselt,
WHO klassifikatsiooni järgi.

Mis on vererõhk?

Vererõhk

Vererõhk on jõud, millega veri surub vastu arteriseinu. Vererõhk on süstoolne, kui süda tõmbub kokku. Vererõhk ondiastoolne, kui süda lõõgastub. Vererõhku mõõdetakse elavhõbeda millimeetrites (mmHg). Inimese loomulik
vererõhk on fundamentaalne rõhk, mida mõõdetakse hommikul kohe pärast ärkamist ja enne söömist, kui keha onveel puhkeasendis.

Mis on hüpertoonia ja kuidas seda kontrollitakse?

Hüpertoonia on ebanormaalselt kõrge arteriaalne vererõhk, millega võivad tähelepanuta jätmise korral kaasneda
terviseprobleemid nagu rabandus ja südameinfarkt. Hüpertoonia kontrolli all hoidmiseks tuleb stressi vältimiseks
elustiili muuta ja võtta arsti järelevalve all rohtu.
Hüpertoonia ennetamiseks või kontrolli all hoidmiseks toimige alljärgnevalt:Ärge suitsetage
Tehke regulaarselt sporti
Tarbige vähem soola ja rasva
Kontrollige regulaarselt tervislikku seisundit
Hoidke kehakaal normis
Eesti 13Miks mõõta vererõhku kodus?

Vererõhu mõõtmine kliinikus või arsti kabinetis võib erutusseisundi tõttu anda kodus mõõdetud vererõhust 25–30
mmHg kõrgema tulemuse. Kodune mõõtmine vähendab välistegurite mõju mõõtmistulemustele, toetab arstimõõtmistulemusi ja annab täpsema ning täielikuma vererõhu ajaloo.
Viide: Journal ofHypertension 1999, köide 17 nr 2

WHO vererõhu klassifikatsioon

Vererõhu muutumine

Inimese vererõhk muutub päeva ja aastaaja lõikes palju. See võibpäeva jooksul erinevate tingimuste tõttu muutuda 30–50 mmHg.
Hüpertooniatõve all kannatavate inimeste puhul on muutused isegi
suuremad. Tavaliselt tõuseb vererõhk töö või mängimise ajal ja
langeb kõige madalamale tasemele magades. Seetõttu ei tasu
eraldiseisva mõõtmistulemuse pärast liialt muret tunda.
Oma normaalse vererõhu teada saamiseks mõõtke vererõhku iga
päev samal ajal, rakendades selles juhendis kirjeldatud protseduure.
Regulaarne mõõtmine annab ülevaatlikuma vererõhu ajaloo.
Vererõhku jäädvustades märkige kindlasti üles kuupäev ja kellaaeg.
Mõõtmistulemuste tõlgendamiseks pidage nõu arstiga.

Eesti

14

2. astme hüpertoonia (mõõdukas)
1. astme hüpertoonia (kerge)Kõrge normaalne
Normaalne
Optimaalne

Süstoolne vererõhk
Tüüpiline kõikumine päeva jooksul(mõõdetud 5-minutiliste intervallidega)
Süstoolne

Vererõhk

Maailma Terviseorganisatsioon (WHO) on välja töötanud
standardid kõrge vererõhu hindamiseks olenemata vanusest(vt tabelit paremal).

Diastoolne vererõhk

3. astme hüpertoonia (raske)

Diastoolne

PM

Magamine

AM

Kellaaeg

Rikkeotsing
Probleem
Ekraanil ei kuvata
midagi isegi pärast
seadmesisselülitamist.

Võimalik põhjus
Patareid on tühjad.
Patarei klemmid ei ole õiges asendis.

Patareides on liiga väike pinge.
Märk(TÜHI PATAREI) vilgub.
Mansett ei täitu õhuga.
Kui patareid on täiesti tühjad, siis seda
märki ei kuvata.
Mansett ei ole korralikult kinnitatud.
Liigutasite mõõtmise ajal oma rannet või
keha.
Seade ei mõõda.
Näidud on kas liiga
Manseti asend ei ole õige.
madalad või liigakõrged.

Muu

Soovituslik tegevus
Asendage kõik patareid uutega.
Paigaldage patareid uuesti, asetades negatiivsed
ja positiivsed klemmid nii, nagu on patareipesas
näidatud.
Asendage kõik patareid uutega.
Kinnitage mansett korralikult.
Püsige mõõtmise ajal täiesti liikumatult ja
vaikselt.
Istuge mugavalt ja ärge liigutage.
Asetage oma käsi pihk ülespoole suunatult lauale
ja mansett südamega samale kõrgusele.
Kui teie südamelöögid on väga nõrgad võiebaregulaarsed, võib seadmel olla raskevererõhku määrata.

Väärtus erineb kliinikus või arsti kabinetis
Lugege jaotist „Miks mõõta vererõhku kodus?”mõõdetust.
Eemaldage patareid. Paigaldage need õigesti
tagasi ja proovige uuesti mõõta.

Märkus. Kui ülalpool kirjeldatud toimingud ei lahenda probleemi, pöörduge edasimüüja poole. Ärge püüdke toodet ise
avada ega parandada, vastasel korral kaotab selle garantii kehtivuse.
Eesti

15

HooldusÄrge avage seadet. See sisaldab õrnu elektrikomponente ja keerukat õhumoodulit, mis võivad kahjustada saada.
Kui probleemi ei õnnestu rikkeotsingu juhiste abil lahendada, pöörduge oma piirkonna volitatud esindaja või meie
klienditeeninduse poole. A&D klienditeenindus pakub volitatud edasimüüjatele tehnilist teavet, varuosi ja seadmeid.
Antud seade on kujundatud ja toodetud nii, et selle kastusiga oleks võimalikult pikk. Sellest hoolimata on soovitatav
lasta seadet iga 2 aasta tagant kontrollida, et tagada selle nõuetekohane töö ja täpsus. Hooldamiseks võtke
ühendust oma piirkonna volitatud esindajaga või ettevõttega A&D.

Tehnilised andmed
TüüpMõõtmismeetodMõõtmisvahemikMõõtmistäpsus
ToiteallikasMõõtmiste arv
Randme ümbermõõt
Klassifikatsioon
Kohaldatav osa
Kasulik tööiga

UB-525
Ostsillomeetriline mõõtmine
Rõhk:
0–299 mmHg
Süstoolne rõhk:60–279 mmHg
Diastoolne rõhk: 40–200 mmHg
Pulss:
40–180 lööki minutis
Rõhk:±3 mmHg
Pulss:±5%
2 x 1,5 V leelispatareid (LR03 või AAA)
Ligikaudu 250 mõõtmist, kui kasutatakse leelispatareisid AAA, rõhk on 170 mmHg,
ruumi temperatuur on 23 °C.
13,5–21,5 cm
Sisemise toitega ME-seade (pidev töörežiim)Manseti tüüp BF
Seade: 5 aastat (kui kasutatakse kuus korda päevas)
Eesti

16

Kliiniline test
EMD
Mälu
Töötingimused
Transpordi-/hoiutingimusedMõõtmed
Kaal
Kaitsetase

Standardi ISO81060-2 : 2013 kohaselt
Kliinilise valideerimise uuringus oli määrati K5 abil 85 uuringualuse diastoolne
vererõhk.
IEC 60601-1-2: 2014
Viimased 60 mõõtmist+10 kuni +40 °C / 15 kuni 85% RH / 800 kuni 1060 hPa-20 kuni +60 °C / 10 kuni 95% RH / 700 kuni 1060 hPa
Ligikaudu 56 [L] x 88 [K] x 21,5 [S] mm
Ligikaudu 90 g, ilma patareideta
IP20

Märkus. Spetsifikatsioone võidakse täiustamiseks ette teatamata muuta.
IP-klassifikatsioon on kaitseaste, mida pakub korpus standardi IEC 60529 kohaselt. Seade on kaitstud
tahkete võõrkehade eest, mille läbimõõt on 12 mm või rohkem (nt sõrmed). Seade ei ole kaitstud vee eest.

Eesti

17

MÄRKUSED

Eesti

18

Saturs

Cienījamie klienti!................................................................................................................................................. 2
Iepriekšējas piezīmes .......................................................................................................................................... 2
Piesardzības pasākumi........................................................................................................................................ 2Sastāvdaļu apzīmējumi........................................................................................................................................ 4
Simboli ................................................................................................................................................................. 5
Mērītāja lietošana ................................................................................................................................................ 7
1. Bateriju ievietošana/nomaiņa ..................................................................................................................... 7
2. Manšetes uzlikšana .................................................................................................................................... 8
3. Kā veikt precīzus mērījumus ...................................................................................................................... 8
4. Mērījums .................................................................................................................................................... 9
5. Pēc mērījuma ............................................................................................................................................. 9
Mērījumi ............................................................................................................................................................. 10Atmiņas datu uzmeklēšana ................................................................................................................................ 11Atmiņā saglabāto datu dzēšana ........................................................................................................................ 11
Kas ir IHB/AFib indikators? ................................................................................................................................ 12
Kas ir priekškambaru fibrilācija? ........................................................................................................................ 12
Spiediena joslas indikators ................................................................................................................................ 12
PVO klasifikācijas indikators .............................................................................................................................. 13
Par asinsspiedienu ............................................................................................................................................ 13
Kas ir asinsspiediens? ................................................................................................................................. 13
Kas ir hipertensija un kā tā tiek kontrolēta?.................................................................................................. 13
Kāpēc nepieciešams mērīt asinsspiedienu mājas apstākļos? ..................................................................... 14
PVO asinsspiediena klasifikācija .................................................................................................................. 14
Asinsspiediena izmaiņas .............................................................................................................................. 14
Problēmu novēršana.......................................................................................................................................... 15
Apkope............................................................................................................................................................... 16
Tehniskie dati..................................................................................................................................................... 16

Latviski1

Cienījamie klienti!

Apsveicam ar modernā A&D asinsspiediena mērītāja iegādi! Šī ierīce, kas paredzēta ērtai lietošanai un precīziem
mērījumiem, atvieglos jūsu ikdienas asinsspiediena mērīšanas režīmu.
Pirms pirmās ierīces lietošanas reizes mēs iesakām rūpīgi izlasīt šo lietošanas rokasgrāmatu.

Iepriekšējas piezīmes
Šī ierīce atbilst Eiropas Padomes Direktīvai 93/42 EEK par medicīnas ierīcēm. Par to liecina
atzīme. (0123: iesaistītās pilnvarotās iestādes atsauces numurs)
Ierīce ir paredzēta lietošanai pieaugušajiem, bet ne jaundzimušajiem vai mazuļiem.
Lietošanas apstākļi. Ierīce ir paredzēta patstāvīgai lietošanai mājas aprūpes apstākļos.
Šī ierīce ir paredzēta, lai noteiktu cilvēka asinsspiedienu un pulsa ātrumu diagnozes uzstādīšanai.

Piesardzības pasākumi

atbilstības

Ierīces konstrukcijā izmantotas precīzijas sastāvdaļas. Nepakļaujiet ierīci paaugstinātai temperatūrai, mitrumam,
tiešiem saules stariem, strāvas triecienam vai putekļiem.
Tīriet ierīci ar sausu, mīkstu drānu vai drānu, kas piesūcināta ar ūdeni un neitrālu mazgāšanas līdzekli. Ierīces
tīrīšanai nekad neizmantojiet spirtu, benzolu, šķīdinātāju vai citas kodīgas ķīmiskās vielas.
Neglabājiet manšeti ilgu laiku cieši savītu, jo tas var saīsināt sastāvdaļu kalpošanas laiku.
Ierīce nav ūdensizturīga. Nepieļaujiet ierīces piesārņošanu ar lietu, sviedriem un ūdeni.
Ja ierīci izmanto televizora, mikroviļņu krāsns, mobilā tālruņa, rentgenstaru vai citu ierīču ar spēcīgu
elektromagnētisko lauku tuvumā, mērījumi var nebūt precīzi.
Lietotās iekārtas, sastāvdaļas un baterijas nav uzskatāmas par parastiem sadzīves atkritumiem, un tās jāiznīcina
saskaņā ar vietējiem noteikumiem.
Atkārtoti lietojot ierīci, pārliecinieties, ka tā ir tīra.
Nepārveidojiet ierīci. Tas var izraisīt negadījumus vai ierīces bojājumus.
Lai mērītu asinsspiedienu, manšetei jāsaspiež plaukstas locītava pietiekoši stipri, lai uz laiku apturētu asins plūsmu
caur artēriju. Tas var izraisīt sāpes vai nejutīgumu plaukstas locītavā, vai arī īslaicīgi izveidot uz plaukstas locītavas
sarkanu plankumu. Šādi simptomi var rasties jo īpaši tad, ja mērījumi tiek secīgi atkārtoti. Pēc brīža sāpes,nejutīgums vai sarkanie plankumi izzudīs.
Šā asinsspiediena mērītāja darbību var ietekmēt arī bezvadu sakaru ierīces, piemēram, mājas tīkla ierīces, mobilie
Latviski2

tālruņi, bezvadu tālruņi, to bāzes stacijas un rācijas. Tāpēc mērītājam jāatrodas vismaz 30 cm attālumā no šādām
ierīcēm.
Ja asinsspiediens tiek mērīts pārāk bieži, asins plūsmas traucējuma dēļ tas var izraisīt traumu. Atkārtoti lietojot
ierīci, pārliecinieties, vai ierīces darbība nerada ilgstošus asinsrites traucējumus.
Klīniskā pārbaude ar zīdaiņiem un grūtniecēm nav veikta. Nelietojiet ierīci zīdaiņiem un grūtniecēm.
Ja jums ir bijusi mastektomija, pirms ierīces lietošanas konsultējieties ar ārstu.
Neļaujiet bērniem pašiem izmantot ierīci un nelietojiet to zīdaiņiem pieejamā vietā. Tas var izraisīt negadījumus vaibojājumus.
Ierīcei ir mazas daļas, kas var radīt aizrīšanās risku, ja zīdainis tās nejauši norij.
Nepieskarieties vienlaikus baterijām un pacientam. Tas var izraisīt elektriskās strāvas triecienu.
Atsevišķu sastāvdaļu bojājumu gadījumā korpuss manšetes tuvumā var uzkarst un izraisīt darbības traucējumus.
Šajā lietošanas rokasgrāmatā neaprakstītu piederumu izmantošana var apdraudēt drošību.
Ja rodas bateriju īssavienojums, tās var sakarst un izraisīt apdegumus.
Pirms ierīces lietošanas ļaujiet tai pielāgoties apkārtējai videi (apmēram vienu stundu).
Nepiepildiet manšeti ar gaisu, ja tā nav aplikta ap plaukstas locītavu.

Kontrindikācijas

Tālāk minēti piesardzības pasākumi pareizai ierīces lietošanai.
Neuzlieciet ierīci uz plaukstas locītavas, ja tai pievienota cita elektriska medicīnas iekārta. Pretējā gadījumā iekārta
var darboties nepareizi.
Cilvēkiem ar smagu asinsrites nepietiekamību rokā pirms ierīces lietošanas jākonsultējas ar ārstu, lai izvairītos no
medicīniskām problēmām.
Pēc mērījuma rezultātu iegūšanas patstāvīgi neuzstādiet diagnozi un nesāciet ārstēšanos. Rezultāti vienmēr
jānovērtē un ārstēšana jānozīmē, konsultējoties ar ārstu.
Nelieciet ierīci uz plaukstas locītavas, uz kuras ir nesadzijusi brūce.
Nelieciet ierīci uz rokas, kurā tiek veikta intravenoza pilināšana vai asins pārliešana. Tas var radīt ievainojumu vai
negadījumus.
Neizmantojiet ierīci vietās, kurās ir viegli uzliesmojošas gāzes, piemēram, anestēzijas gāzes. Tas var izraisīt
eksploziju.
Neizmantojiet ierīci ļoti koncentrēta skābekļa vidē, piemēram, augstspiediena skābekļa kamerā vai skābekļa teltī.
Tas var izraisīt degšanu vai sprādzienu.
Latviski3

Sastāvdaļu apzīmējumi
Displejs

poga (vidējais, atmiņa un iepriekš iestatīto parametru nomaiņa)START (SĀKT) poga

Bateriju nodalījuma vāks

ManšeteATMIŅA
Vidējs

Displejs

Sārma baterijas (LR03 vai AAA)

Sistoliskais spiediens
Diastoliskais spiediens

PVO klasifikācijas indikators
un
Spiediena joslas indikators

IHB/AFib simbols

Sirds atzīme

Pulsa ātrums

Bateriju indikators
Latviski4

Simboli

Simboli, kas uzdrukāti uz ierīces korpusa
Simboli

Funkcija/nozīme

Gaidstāve un ierīces ieslēgšana
SYS

Sistoliskais asinsspiediens, kas izteikts mm Hg

DIA

Diastoliskais asinsspiediens, kas izteikts mm Hg

PUL

Pulss minūtē
Bateriju ielikšanas norādījums
Līdzstrāva

SN

Sērijas numursRažošanas datums
BF tips: ierīce un manšete ir izstrādātas tā, lai nodrošinātu īpašu aizsardzību pret elektriskās strāvas
triecieniem.
EK direktīvas medicīnas ierīces uzlīme

IP

Starptautiskais aizsardzības simbols
WEEE uzlīmeRažotājs
ES pārstāvis
Skatiet lietošanas rokasgrāmatu/brošūru
Glabāt sausā vietā
Latviski5

Simboli, kas tiek attēloti uz displeja
Simboli

Funkcija/nozīme/ieteicamā rīcība
Parādās mērījuma veikšanas laikā. Mirgo, kad ir noteikts pulss.
Mēģiniet nekustēties.
IHB/AFib simbols parādās, kad ir konstatēta neregulāra sirdsdarbība.
Var iedegties, ja tiek konstatēta neliela vibrācija, piemēram, trīce vai kratīšana.

Simboli, kas tiek attēloti uz displeja (turpinājums)
Simbolivai

Funkcija/nozīme
Iepriekšējie mērījumi, kas saglabāti ATMIŅĀ
Vidējie dati
PILNA BATERIJA
Bateriju uzlādes indikators mērījuma laikā
ZEMS BATERIJAS UZLĀDES LĪMENIS
Ja indikators mirgo, baterijas uzlādes līmenis ir zems

Ieteicamā rīcība

Kad indikators mirgo, aizstājiet visas
baterijas ar jaunām.
Veiciet citu mērījumu.
Nestabils asinsspiediens, ko izraisa kustēšanās mērījuma laikā.
Mērījuma laikā nekustieties.
Sistoliskās un diastoliskās vērtības ir 10 mm Hg viena pret otru.
Gaisa piepūšanas laikā spiediena vērtība nepalielinājās.
Uzlieciet manšeti pareizi un vēlreiz
Manšete nav pareizi uzlikta.
veiciet mērījumu.
PULSA DISPLEJA KĻŪDA
Pulss netika pareizi noteikts.
Izņemiet baterijas un nospiediet
pogu START (SĀKT), tad vēlreiz
Asinsspiediena mērītāja iekšējā kļūda
ievietojiet baterijas. Ja kļūda parādās
vēlreiz, sazinieties ar tirgotāju.
Latviski6

1. Bateriju ievietošana/nomaiņa

Mērītāja lietošana

1. Noņemiet bateriju nodalījuma vāku.

Bateriju
nodalījuma vāks

2. Izņemiet izlietotās baterijas un bateriju nodalījumā ievietojiet
jaunas baterijas, kā parādīts attēlā, lai polaritātes (+ un -)būtu pareizas. Izmantojiet tikai LR03 vai AAA baterijas.
3. Uzlieciet atpakaļ bateriju nodalījuma vāku.

Izlietotās baterijas

UZMANĪBU
Ievietojiet baterijas, kā parādīts uz bateriju nodalījuma.
Jaunās baterijas
Ja tās tiek ievietotas nepareizi, ierīce nedarbosies.
Kad uz displeja mirgo(atzīme ZEMS BATERIJAS
UZLĀDES LĪMENIS), aizstājiet visas baterijas ar jaunām.
Nesajauciet vecās baterijas ar jaunajām.Tas var samazinātbateriju ekspluatācijas laiku vai kavēt ierīces darbību.
(atzīme ZEMS BATERIJAS UZLĀDES LĪMENIS)
neparādās, kad baterijas ir tukšas.
Bateriju ekspluatācijas laiks ir atkarīgs no apkārtējās vides temperatūras, un zemās temperatūrās tas var būt īsāks.
Parasti divu jaunu LR03/AAA bateriju ekspluatācijas laiks ir apmēram četri mēneši, ja tās izmanto mērījumam divas
reizes katru dienu.
Izmantojiet tikai norādītās baterijas. Baterijas, kas ir iekļautas komplektā ar ierīci, ir paredzētas, lai pārbaudītu
mērītāja darbību, un to ekspluatācijas laiks ir ierobežots.
Ja ierīce ilgāku laika periodu netiks lietota, izņemiet baterijas. Baterijām var rasties noplūde, tādējādi izraisot
darbības traucējumu.

Latviski7

2. Manšetes uzlikšana
1. Aptiniet manšeti ap plaukstas locītavu aptuveni 1 cm attālumā no plaukstas, kāparādīts attēlā pa labi.

Aptuveni 1 cm

2. Cieši nostipriniet manšeti ar līpaizdares sloksni.
Piezīme.

Lai mērījumi būtu precīzi, uzlieciet manšeti cieši un veiciet mērījumu uz
atsegtas plaukstas locītavas.

3. Kā veikt precīzus mērījumus

Lai veiktu pēc iespējas precīzāku asinsspiediena mērījumu:
Mērījuma laikā nekustieties un klusējiet.
Ērti apsēdieties uz krēsla. Nolieciet elkoni uz galda ar plaukstu vērstu augšup, lai manšete atrastos sirds līmenī.
Pirms mērījuma atslābinieties piecas līdz desmit minūtes. Ja jūs esat satraukts vai nomākts emocionāla stresa dēļ,šī stresa ietekmē asinsspiediena rādījums būs augstāks (vai zemāks) par parasto, un pulsa rādījums parasti irātrāks nekā parasti.
Centieties veikt asinsspiediena mērījumu katru dienu aptuveni vienā un tajā pašā laikā.
Asinsspiediens pastāvīgi mainās atkarībā no tā, ko cilvēks dara un ko ir ēdis vai dzēris — tam var būt ļoti spēcīga
un strauja ietekme uz asinsspiedienu.
Neveiciet mērījumu uzreiz pēc fiziskas slodzes vai mazgāšanās vannā. Pirms mērījuma veikšanas atpūtieties
divdesmit vai trīsdesmit minūtes.
Nesakrustojiet kājas. Nolieciet pēdas līdzeni uz grīdas un iztaisnojiet muguru.
Ierīces mērījumu pamatā ir sirdsdarbība. Ja jums ir ļoti vāja vai neregulāra sirdsdarbība, ierīcei var būt grūti noteikt
asinsspiedienu.
Ja ierīce konstatē neparastu stāvokli, tā pārtrauc mērījumu un parāda kļūdas simbolu. Simbolu aprakstu skatiet
6. lpp.
Šo asinsspiediena mērītāju drīkst lietot pieaugušais. Pirms šīs ierīces lietošanas bērnam konsultējieties ar savuārstu. Bērns nedrīkst lietot šo ierīci bez uzraudzības.
Automātiskā asinsspiediena mērītāja darbību var ietekmēt pārmērīga temperatūra, mitrums vai augstums.
Latviski8

4. Mērījums

Mērījuma laikā manšete kļūst ļoti cieša — tā ir normāla parādība.

5. Pēc mērījuma

Ja tiek nospiesta poga START (SĀKT), lai izslēgtu ierīci, kamēr ir redzami rādījumi, jaunie rādījumi tiek saglabātiatmiņā.
Ja tiek nospiesta poga , lai izslēgtu ierīci, kamēr ir redzami rādījumi, jaunie rādījumi netiek saglabāti.
Noņemiet manšeti un ierakstiet savus datus.
Piezīmes.

Ierīce ir aprīkota ar automātiskas strāvas padeves izslēgšanas funkciju, kas saglabā pašreizējos datusatmiņā un izslēdz ierīci aptuveni vienu minūti pēc mērīšanas. Starp mērījumiem vienai un tai pašai
personai jābūt vismaz trīs minūšu starplaikam.

Latviski

9

Mērījumi
Piezīme. Pēc pirmās lietošanas reizes UB-525 nodrošina lietotājam atbilstošu gaisa
piepūšanas spiedienu.
1. Aplieciet manšeti ap plaukstas locītavu. Apsēdieties ērti, novietojiet manšeti
vienā līmenī ar jūsu sirdi un atslābinieties.
2. Nospiediet pogu START (SĀKT). Redzami visi displeja segmenti.

Visi segmenti

3. Uz īsu brīdi mirgo nulles atzīme (0). Kad sākas mērījums, tad displejs mainās.
Manšete sāk piepildīties ar gaisu. Manšete kļūst ļoti cieša — tā ir normāla
parādība. Mērījums tiek sākts automātiski, kad sākas gaisa piepūšana, un
mirgo(sirds atzīme).

Nulles rādījums
Gaisa piepūšana ir sākta

Piezīme. Ja jebkurā brīdī vēlaties apturēt gaisa piepūšanu, vēlreiz nospiediet
pogu START (SĀKT).

Tiek veikta piepildīšana,un notiek mērījums

4. Kad mērījums ir pabeigts, ierīcē tiek attēloti rādījumi (sistoliskā un diastoliskā
spiediena rādījumi, pulsa ātrums, PVO klasifikācija, neregulāras sirdsdarbības
simbols). Manšete automātiski izlaiž atlikušo gaisu un pilnībā iztukšojas.
Piezīme. Ja nevēlaties saglabāt jaunos rādījumus atmiņā, nospiediet pogu
kamēr rādījumi ir redzami.
5. Lai izslēgtu ierīci, vēlreiz nospiediet pogu START (SĀKT). Noņemiet
manšeti.
Piezīmes. Ierīce ir aprīkota ar automātiskas strāvas padeves izslēgšanas
funkciju.
Starp mērījumiem vienai un tai pašai personai jābūt vismaz trīs
minūšu starplaikam.

Latviski

10,

Sistoliskais spiediens
Diastoliskais spiediens
PVO klasifikācija

Neregulāras sirdsdarbības simbols
Pulsa ātrumsAtmiņas datu uzmeklēšana
Piezīme.
Šī ierīce saglabā atmiņā pēdējos 60 mērījumus.
1. Izslēdzot ierīci, nospiediet pogu .
Tiek parādīta visu mērījumu vidējā vērtība un datu skaits. Ja datu nav,tiek rādīts “0”. Lai izslēgtu ierīci, nospiediet pogu
vai START (SĀKT).
2. Izmantojiet tālāk norādītās pogas, lai attēlotu datus (skaitu un mērījumu
datus).
Ierīcē tiek parādīta visu mērījumu vidējā vērtība un datu skaits.
Katru reizi nospiežot pogu , ierīce parāda datus šādā secībā:visu starplaikā veikto mērījumu vidējie dati;
dati (skaits un mērījumu dati).
Ierīcē dati tiek rādīti, sākot no jaunākajiem.
Piemērā:Nr. 30 un dati
Nr. 29 un dati
Nr. 01 un dati.
pēc tam, kad tiek parādīti senākie dati, ierīce
3. Ja nospiežat pogupāriet uz 1. darbību, un tiek parādīta visu mērījumu vidējā vērtība un
datu skaits.
4. Lai izslēgtu ierīci, nospiediet pogu START (SĀKT).
Ja ar ierīci vienu minūti neveic nekādas darbības, tā automātiski izslēdzas.

Atmiņas režīms

poga
Visu datu vidējais
rādītājs
Visu datu skaits
Piem.: Nr. 30
Pēdējā mērījuma
datiAtmiņā saglabāto datu dzēšana
1. Nospiediet un turiet pogu

, līdz ierīce automātiski izslēdzas.

, atmiņā saglabātie dati tiek dzēsti, kamēr atzīme
Ierīcē tiek parādīta atzīme
mirgo, un ierīce automātiski izslēdzas.
Latviski

11

Izslēgšana

poga

poga

Pirmā mērījuma
dati (vecākā)poga
IzslēgšanapogaKas ir IHB/AFib indikators?
Kad mērījuma laikā mērītājs konstatē neregulāru ritmu, uz displeja parādās IHB/AFib indikators ar mērījuma
vērtībām.
Piezīme. Ja šis

IHB/AFib indikators parādās bieži, mēs iesakām sazināties ar savu ārstu.

Kas ir priekškambaru fibrilācija?

Sirds saraujas, sirdī rodoties elektriskiem impulsiem, un sūknē asinis caur ķermeni. Priekškambaru fibrilācija (AFib)rodas, kad elektriskie impulsi priekškambarī ir traucēti, tādējādi izraisot neregulāru pulsa intervālu. Priekškambaru
fibrilācija var izraisīt asiņu stagnāciju sirdī, kas var viegli radīt asins recekļus — triekas vai sirdslēkmes cēloni.

Spiediena joslas indikators

Palielinās spiediens

Indikators uzrauga spiediena palielināšanos vai
samazināšanos mērījuma laikā.
Notiek gaisa piepūšana
Notiek mērījums
Gaisa piepūšana pabeigta

Latviski

12

PVO klasifikācijas indikators
Katrs no sešiem joslas indikatora segmentiem atbilst PVO
asinsspiediena klasifikācijai, kā aprakstīts 14. lpp.
Piemērs
Mērena hipertensija Viegla hipertensija

Augsts normāls

PVO klasifikācijas indikators
Smaga hipertensija
Mērena hipertensija
Viegla hipertensija
Augsts normāls
Normāls
Optimāls
: Indikators uzrāda segmentu,
pamatojoties uz pašreizējiem datiem,
kas atbilst PVO klasifikācijai.

Kas ir asinsspiediens?

Par asinsspiedienu

Asinsspiediens ir spēks, ar kādu asinis spiež uz artēriju sieniņām. Sistoliskais spiediens rodas, sirdij saraujoties.
Diastoliskais spiediens rodas, sirdij izplešoties. Asinsspiediens tiek mērīts dzīvsudraba stabiņa milimetros (mm Hg).
Cilvēka dabiskais asinsspiediens tiek atspoguļots pamatspiediena ietekmē, ko mēra no rīta pēc pamošanās, kadcilvēks vēl ir miera stāvoklī un nav ēdis.

Kas ir hipertensija un kā tā tiek kontrolēta?

Ja hipertensiju jeb patoloģiski augstu arteriālo asinsspiedienu neārstē, tā var izraisīt daudzas veselības problēmas,tostarp insultu un sirdslēkmi. Hipertensiju var kontrolēt, mainot dzīvesveidu, izvairoties no stresa un lietojot zālesārsta uzraudzībā.
Lai novērstu hipertensiju vai to kontrolētu:
nesmēķējiet;
regulāri vingrojiet;
samaziniet uzņemtā sāls un tauku daudzumu;
regulāri veiciet veselības pārbaudes;
uzturiet pareizu svaru.
Latviski13

Kāpēc nepieciešams mērīt asinsspiedienu mājas apstākļos?

Pasaules Veselības organizācija (PVO) ir noteikusi augsta
asinsspiediena novērtēšanas standartus neatkarīgi no vecuma,
kā parādīts diagrammā pa labi.

Asinsspiediena izmaiņas

Asinsspiediens ievērojami atšķiras atkarībā no dienas un gadalaika.
Dažādu apstākļu dēļ dienas laikā tas var svārstīties no 30 līdz
50 mm Hg. Pacientiem ar hipertensiju izmaiņas ir vēl izteiktākas.
Parasti asinsspiediens paaugstinās, strādājot vai spēlējoties, unmiega laikā samazinās līdz zemākajam līmenim. Tāpēcneuztraucieties par viena mērījuma rezultātiem.
Lai noteiktu savu normālo asinsspiedienu, veiciet mērījumus katru
dienu vienā un tajā pašā laikā, izmantojot šajā lietošanas
rokasgrāmatā aprakstīto procedūru. Regulāri rādījumi sniedz
daudzpusīgāku asinsspiediena datu vēsturi. Ierakstot
asinsspiedienu, noteikti norādiet datumu un laiku. Konsultējieties arsavu ārstu, lai novērtētu asinsspiediena datus.

Latviski

14

Hypertension” 1999, Vol 17 No.2
3. pakāpes hipertensija
2. pakāpes hipertensija (mērena)
1. pakāpes hipertensija (viegla)
Augsta, normāla
Normāls
Optimāls

Sistoliskais asinsspiediens
Tipiskas svārstības vienas dienas laikā(mērīts pēc katrām piecām minūtēm)SistoleAsinsspiediens

PVO asinsspiediena klasifikācija

Diastoliskais asinsspiediensAsinsspiediena mērīšana klīnikā vai ārsta kabinetā var izraisīt trauksmi, tādējādi radot paaugstinātu rādījumu, kas ir
par 25 līdz 30 mm Hg lielāks nekā par mājās izmērīto. Mērījums mājās samazina ārējo apstākļu ietekmi uz
asinsspiediena rādījumiem, papildina ārsta rādījumus, kā arī nodrošina precīzāku un pilnīgāku asinsspiediena datu
vēsturi.
Atsauces materiāls: “Journal of

Diastole
Pēc pusdienlaika

Miegs
Pirms pusdienlaika Laiks

Problēmu novēršana
Problēma
Displejā nekas
neparādās, pat ja
ierīce ir ieslēgta.

Manšete neuzpildās.

Ierīce nemēra.
Rādījumi ir pārākaugsti vai zemi.

Cita problēma

Iespējamais cēlonis
Baterijas ir tukšas.

Ieteicamā rīcībaAizstājiet visas baterijas ar jaunām.
Atkārtoti ievietojiet baterijas tā, lai negatīvie unpozitīvie kontakti atbilstu bateriju nodalījumā
norādītajiem.

Bateriju kontakti nav pareizi novietoti.
Pārāk zems bateriju spriegums.
Mirgo(atzīme ZEMS BATERIJAS
UZLĀDES LĪMENIS).
Atzīme neparādās, ja bateriju
ekspluatācijas laiks ir pilnībā beidzies.
Manšete nav pareizi uzlikta.
Mērījuma laikā jūs pakustinājāt plaukstas
locītavu vai ķermeni.
Manšetes novietojums nav pareizs.

Vērtība atšķiras no klīnikā vai ārsta
kabinetā noteiktās.

Aizstājiet visas baterijas ar jaunām.

Uzlieciet manšeti pareizi.
Centieties mērījuma laikā pilnībā nekustēties unnerunāt.
Apsēdieties ērti un nekustieties.
Nolieciet roku uz galda ar plaukstu vērstu
augšup, un uzlieciet manšeti sirds līmenī.
Ja jums ir ļoti vāja vai neregulāra sirdsdarbība,
ierīcei var būt grūti noteikt jūsu asinsspiedienu.
Skatiet sadaļu “Kāpēc nepieciešams mērīt
asinsspiedienu mājas apstākļos?”.
Izņemiet baterijas. Ievietojiet tās pareizi unmēģiniet mērīt vēlreiz.

Piezīme. Ja iepriekš aprakstītās darbības neatrisina problēmu, sazinieties ar tirgotāju. Nemēģiniet atvērt vai labot šoizstrādājumu saviem spēkiem, jo pretējā gadījumā jūsu garantija būs nederīga.
Latviski15

Apkope
Neatveriet ierīci. Tajā ietvertas trauslas elektriskās daļas un sarežģīts gaisa bloks, kas var tikt bojāts. Ja nevarat
novērst problēmu, izmantojot problēmu novēršanas norādījumus, sazinieties ar pilnvaroto tirgotāju savā reģionā vai
mūsu klientu apkalpošanas nodaļu. A&D klientu apkalpošana nodrošinās tehnisko informāciju, rezerves daļas un
blokus pilnvarotajiem tirgotājiem.
Ierīce tika izstrādāta un izgatavota ilgstošam ekspluatācijas laikam. Tomēr parasti ierīci ir ieteicams pārbaudīt ik
pēc 2 gadiem, lai nodrošinātu pareizu un precīzu darbību. Lai veiktu apkopi, lūdzu, sazinieties ar pilnvarotu tirgotāju
jūsu reģionā vai uzņēmumu A&D.

Tehniskie dati
Tips
Mērījuma metodeUB-525

Oscilometrisks mērījums
Spiediens:0 – 299 mm Hg
Sistoliskais spiediens:
60 – 279 mm Hg
Mērījuma diapazons
Diastoliskais spiediens:
40 – 200 mm Hg
Pulss:
40 – 180 sitieni/minūtē
Spiediens: ±3 mm Hg
Mērījuma precizitāte
Pulss:±5 %Strāvas padeve2 x 1,5 V sārma baterijas (LR03 vai AAA)
Aptuveni 250 mērījumu, ja tiek izmantotas AAA sārma baterijas, ar spiediena vērtību
Mērījumu skaits
170 mm Hg istabas temperatūrā 23 °C.
Plaukstas locītavas apkārtmērs 13,5–21,5 cm
Klasifikācija
Medicīnas elektroiekārta ar iekšēju barošanas avotu (nepārtraukts darbības režīms)Daļa, kas saskaras ar pacientu Manšetes tips BF
Lietderīgās lietošanas laiks
Ierīce: 5 gadi (ja lieto sešas reizes dienā)
Latviski

16

Klīniskā pārbaude
Elektromagnētiskie traucējumi
AtmiņaDarbības apstākļi
Piegādes/uzglabāšanas
apstākļi
Izmēri
Svars
Aizsardzība pret iekļūšanu
Piezīme.

Saskaņā ar ISO81060-2：2013
Klīniskās validācijas pētījumā K5 tika izmantots 85 subjektiem, lai noteiktu diastolisko
asinsspiedienu.
IEC 60601-1-2: 2014
Pēdējie 60 mērījumi
No +10 līdz +40 °C/no 15 līdz 85% RH/no 800 līdz 1060 hPa
No -20 līdz +60 °C/no 10 līdz 95% RH/no 700 līdz 1060 hPa
Aptuveni 56 [P] x 88 [A] x 21,5 [D] mm
Aptuveni 90 g bez baterijām
IP20Specifikācijas uzlabošanas nolūkos var tikt mainītas bez iepriekšēja brīdinājuma.
IP klasifikācija ir aizsardzības pakāpe, ko nodrošina pielikumi saskaņā ar IEC 60529. Šī ierīce ir aizsargāta
pret cietiem svešķermeņiem, kuru diametrs ir 12 mm vai lielāks, piemēram, pirkstiem. Ierīce nav aizsargāta
pret ūdens iedarbību.

Latviski17

PIEZĪMES

Latviski18

TurinysGerb. klientai........................................................................................................................................................ 2Įžanginės pastabos .............................................................................................................................................. 2
Atsargumo priemonės .......................................................................................................................................... 2Dalių identifikavimas ............................................................................................................................................ 4
Simboliai .............................................................................................................................................................. 5
Kaip naudoti matuoklį .......................................................................................................................................... 7
1. Kaip įdėti arba pakeisti maitinimo elementus ............................................................................................. 7
2. Kaip užsidėti rankovę ................................................................................................................................. 8
3. Kaip atlikti tikslius matavimus ..................................................................................................................... 8
4. Matavimas .................................................................................................................................................. 9
5. Po matavimo .............................................................................................................................................. 9
Matavimai .......................................................................................................................................................... 10
Atminties duomenų atkūrimas ........................................................................................................................... 11
Kaip pašalinti atmintyje saugomus duomenis .................................................................................................... 11
Kas yra IHB/AFib indikatorius? .......................................................................................................................... 12
Kas yra AFib? .................................................................................................................................................... 12
Spaudimo juostos indikatorius ........................................................................................................................... 12
PSO klasifikavimo indikatorius ........................................................................................................................... 13
Apie kraujospūdį ................................................................................................................................................ 13
Kas yra kraujospūdis? .................................................................................................................................. 13
Kas yra hipertenzija ir kaip ji kontroliuojama? .............................................................................................. 13Kodėl verta matuoti kraujospūdį namuose? ................................................................................................. 14
PSO kraujospūdžio klasifikacija ................................................................................................................... 14
Kraujospūdžio pokyčiai ................................................................................................................................ 14
Trikčių šalinimas ................................................................................................................................................ 15
Techninė priežiūra ............................................................................................................................................. 16
Techniniai duomenys ......................................................................................................................................... 16

Lietuvių k.

1

Gerb. klientai
Sveikiname įsigijus naujausią „A&D“ kraujospūdžio matuoklį. Sukurtas taip, kad būtų paprastas naudoti ir tikslus,
šis prietaisas padės kasdien stebėti kraujospūdį.
Prieš pirmą kartą naudojant prietaisą, rekomenduojame atidžiai perskaityti šį vadovą.

Įžanginės pastabosŠis prietaisas atitinka Europos direktyvą 93/42 EEB dėl medicinos prietaisų. Tai įrodo atitikties simbolis(0123 yra atitinkamos notifikuotosios įstaigos nuorodos numeris.)
Prietaisas skirtas naudoti suaugusiesiems, ne naujagimiams ar kūdikiams.
Naudojimo aplinka. Prietaisas skirtas naudoti ne specialistams namų aplinkoje.
Šis prietaisas sukurtas diagnozavimo tikslais matuoti žmogaus kraujospūdį ir pulsą.

.

Atsargumo priemonės
Šiame prietaise yra itin tikslių komponentų. Reikia vengti didelių temperatūros ir drėgmės svyravimų, tiesioginės
saulės šviesos, smūgių ir dulkių.
Prietaisą valykite sausu švelniu audiniu arba audiniu, sudrėkintu vandeniu ir neutraliu plovikliu. Prietaisui valytiniekada nenaudokite alkoholio, benzeno, skiediklio ar kitų stiprių chemikalų.
Pasistenkite ilgai nelaikyti rankovės tvirtai sulankstytos, nes tai gali sutrumpinti jos naudojimo laikotarpį.
Prietaisas neatsparus vandeniui. Pasistenkite, kad lietus, prakaitas ar vanduo nesuteptų prietaiso.
Jei prietaisas naudojamas greta televizoriaus, mikrobangų krosnelės, mobiliųjų telefonų, rentgeno ar kitų įrenginių,skleidžiančių stiprius elektrinius laukus, matavimo rezultatai gali būti iškraipyti.
Panaudota įranga, dalys ir maitinimo elementai nėra laikomi įprastomis buitinėmis atliekomis, juos reikia išmesti
pagal atitinkamas vietos taisykles.
Naudodami pakartotinai įsitikinkite, kad prietaisas yra švarus.
Nemodifikuokite prietaiso. Tai gali tapti nelaimingų atsitikimų priežastimi ar sugadinti prietaisą.
Norint matuoti kraujospūdį, rankovė turi suspausti riešą pakankamai stipriai, kad kraujo tekėjimas arterija būtųlaikinai sustabdytas. Tai gali sukelti rankos skausmą, tirpimą arba laikiną paraudimą. Tai ypač dažnai pasitaiko, kaimatavimai atliekami vienas paskui kitą. Bet koks skausmas, tirpimas arba paraudimas pradings bėgant laikui.
Lietuvių k.

2

Belaidžiai ryšio prietaisai, pvz., namuose naudojami tinklo prietaisai, mobilieji telefonai, belaidžiai telefonai ir jųbazinės stotys, portatyvios radijo stotelės, gali paveikti šį kraujospūdžio matuoklį. Todėl nuo tokių prietaisų reikia
laikytis mažiausiai 30 cm atstumo.
Pernelyg dažnas kraujospūdžio matavimas gali pakenkti, nes trikdoma kraujo apytaka. Pakartotinai naudodami
prietaisą įsitikinkite, kad prietaiso naudojimas pernelyg ilgai netrikdo kraujotakos.
Klinikiniai tyrimai su naujagimiais ir nėščiomis moterimis nebuvo atlikti. Nenaudokite naujagimiams ar nėščioms
moterims.
Jeigu jums atliktas krūties pašalinimas, prieš naudodama prietaisą pasitarkite su gydytoju.
Neleiskite vaikams patiems naudotis prietaisu ir nenaudokite prietaiso kūdikiams pasiekiamoje vietoje. Tai gali tapti
nelaimingų atsitikimų priežastimi.
Yra smulkių dalių, galinčių kelti užspringimo pavojų, jei jas per klaidą prarytų kūdikiai.
Tuo pačiu metu nelieskite maitinimo elementų ir paciento. Tai gali sukelti elektros smūgį.
Sutrikus vieno komponento veikimui, korpusas šalia rankovės gali įkaisti ir prietaiso veikimas gali sutrikti.
Šiame vadove nenurodytų priedų naudojimas gali pakenkti saugai.
Jei įvyksta trumpasis maitinimo elemento jungimas, jis gali įkaisti ir nudeginti.
Prieš naudodami leiskite prietaisui prisitaikyti prie supančios aplinkos (maždaug vieną valandą).
Pradėkite pūsti tik uždėję rankovę ant riešo.

Kontraindikacijos

Toliau nurodomos atsargumo priemonės, užtikrinančios tinkamą prietaiso naudojimą.
Nedėkite prietaiso ant riešo, jei prijungta kita elektrinė medicinos įranga. Įranga gali veikti netinkamai.
Asmenys, turintys rimtą rankos kraujotakos deficitą, prieš naudodami prietaisą turi pasitarti su gydytoju, kadišvengtų sveikatos sutrikimų.
Nevertinkite matavimo rezultatų ir nepradėkite gydymo savarankiškai. Visada pasitarkite su savo gydytoju dėl
rezultatų vertinimo ir gydymo.
Nedėkite prietaiso ant riešo, jei ant jo yra neužgijusi žaizda.
Nedėkite prietaiso ant rankos, jei prijungta intraveninė lašinė ar vykdomas kraujo perpylimas. Tai gali taptisužeidimo ar nelaimingų atsitikimų priežastimi.
Nenaudokite prietaiso aplinkoje, kurioje yra degių dujų, pvz., anestezinių dujų. Tai gali sukelti sprogimą.
Nenaudokite prietaiso aplinkoje, kurioje yra didelė deguonies koncentracija, pvz., didelio slėgio deguonies kameroje
arba deguonies palapinėje. Tai gali sukelti gaisrą arba sprogimą.
Lietuvių k.

3

Dalių identifikavimas
EkranasMygtukas(vidurkis, atmintis ir iš anksto nustatytų parametrų keitimas)

START mygtukas

Maitinimo elementų
dangtelis

RankovėATMINTIS
Vidurkis

EkranasŠarminiai maitinimo elementai (LR03 arba AAA)

Sistolinis spaudimas
Diastolinis spaudimas

PSO klasifikavimo indikatorius
ir
spaudimo juostos indikatorius

IHB/AFib simbolisŠirdies simbolis

Pulso dažnis

Maitinimo elementų indikatorius

Lietuvių k.

4

Simboliai

Simboliai, pavaizduoti ant prietaiso dėžutės
Simboliai

Funkcija / reikšmė

Budėjimo režimas ir prietaiso įjungimas.
SYS

Sistolinis kraujospūdis (mmHg)

DIA

Diastolinis kraujospūdis (mmHg)

PUL

Pulsas per minutę
Maitinimo elementų įdėjimo orientyras
Nuolatinė srovė

SN

Serijos numeris
Pagaminimo data
BF tipas: prietaisas ir rankovė yra sukurti taip, kad suteiktų specialią apsaugą nuo elektros smūgių.
EB direktyvos medicinos prietaisų etiketė

IP

Tarptautinis apsaugos simbolisEEĮA etiketė
Gamintojas
ES atstovas
Žr. naudotojo vadovą / lankstinuką.
Laikyti sausai.

Lietuvių k.

5

Ekrane rodomi simboliai
Simboliai

Funkcija / reikšmė / rekomenduojamas veiksmas
Rodomas, kai vyksta matavimas. Mirksi, kai aptinkamas pulsas.
Pasistenkite nejudėti.
IHB/AFib simbolis rodomas, kai aptinkamas nereguliarus širdies ritmas.
Gali įsižiebti, kai aptinkamas labai nežymus vibravimas, pvz., virpėjimas arba drebėjimas.

Ekrane rodomi simboliai (tęsinys)
Simboliai

Funkcija / reikšmė
Ankstesni matavimo rezultatai, išsaugoti ATMINTYJE
Duomenų vidurkis
MAITINIMO ELEMENTAI VISIŠKAI ĮKRAUTI
Maitinimo elementų indikatorius matuojant
MAITINIMO ELEMENTAI SENKA
Kai mirksi, maitinimo elementai senka

Nestabilus kraujospūdis dėl judėjimo matuojant
arba

Rekomenduojamas veiksmas

Kai indikatorius mirksi, pakeiskite
visus maitinimo elementus naujais.
Dar kartą matuokite kraujospūdį.
Matuodami kraujospūdį nejudėkite.

Sistolinės ir diastolinės reikšmės skiriasi ne daugiau kaip
10 mmHg
Pripučiant slėgio reikšmė nepadidėjo
Rankovė uždėta netinkamai
PUL. RODYMO EKRANE KLAIDA
Pulsas tinkamai neaptiktas

Tinkamai uždėkite rankovę ir dar kartą
matuokite kraujospūdį.

Kraujospūdžio matuoklio vidinė klaida

Išimkite maitinimo elementus ir
spauskite mygtuką START , tuometvėl įdėkite maitinimo elementus. Jei
klaida vis tiek rodoma, susisiekite su
pardavėju.
Lietuvių k.

6

Kaip naudoti matuoklį

1. Kaip įdėti arba pakeisti maitinimo elementus
1. Nuimkite maitinimo elementų dangtelį.

Maitinimo elementų
dangtelis

2. Išimkite panaudotus maitinimo elementus ir, kaip pavaizduota,į jų skyrelį įdėkite naujus pasirūpindami, kad poliai (+ ir –)būtų tinkami. Naudokite tik LR03 arba AAA maitinimo
elementus.
3. Uždėkite maitinimo elementų dangtelį.

Panaudoti maitinimo
elementai

PERSPĖJIMAI

Nauji maitinimo elementaiĮdėkite maitinimo elementus į jų skyrelį, kaip pavaizduota.
Jei jie įdėti netinkamai, prietaisas neveiks.
Kai ekrane mirksi(simbolis MAITINIMO ELEMENTAISENKA), visus maitinimo elementus pakeiskite naujais.
Nenaudokite senų maitinimo elementų kartu su naujais. Tai
gali sutrumpinti maitinimo elementų naudojimo laikotarpį arba
prietaisas gali veikti netinkamai.
(simbolis MAITINIMO ELEMENTAI SENKA) nerodomas,
kai maitinimo elementai yra išsikrovę.
Maitinimo elementų naudojimo laikotarpis priklauso nuo aplinkos temperatūros ir gali sutrumpėti esant žemai
temperatūrai. Paprastai du nauji LR03 / AAA maitinimo elementai naudojami maždaug keturis mėnesius, jei
kiekvieną dieną atliekami du kraujospūdžio matavimai.
Naudokite tik nurodytus maitinimo elementus. Su prietaisu pateikiami maitinimo elementai skirti išbandyti matuoklio
veikimą, jų naudojimo laikotarpis gali būti ribotas.
Išimkite maitinimo elementus, jei prietaiso nenaudosite ilgą laiką. Galimas maitinimo elementų nuotėkis, dėl kurio
prietaisas gali pradėti netinkamai veikti.
Lietuvių k.

7

2. Kaip užsidėti rankovę
1. Apsukite rankovę aplink riešą taip, kad iki plaštakos liktų maždaug 1 cm, kaip
parodyta paveikslėlyje dešinėje.

Maždaug 1 cm

2. Pritvirtinkite rankovę naudodami kibiąją juostelę.
Pastaba:

kad matavimas būtų tikslus, rankovė turi gerai priglusti prie pliko riešo.

3. Kaip atlikti tikslius matavimusNorėdami užtikrinti itin tikslius kraujospūdžio matavimo rezultatus, atlikite nurodytus veiksmus.
Atlikdami matavimą nejudėkite ir nekalbėkite.
Patogiai atsisėskite. Padėkite alkūnę ant stalo, pasukite delną į viršų ir patikrinkite, ar rankovė yra širdies lygyje.
Prieš pradėdami matuoti kraujospūdį pabūkite ramybės būsenoje nuo penkių iki dešimties minučių. Jei esate
susijaudinę ar jaučiate emocinį stresą, matavimas parodys šį stresą kaip aukštesnį (arba žemesnį) nei įprasta
kraujospūdį, o pulsas bus greitesnis nei įprasta.
Pasistenkite kraujospūdį matuoti kiekvieną dieną maždaug tuo pačiu metu.
Žmogaus kraujospūdis nuolat kinta ir priklauso nuo veiklos; maistas bei gėrimai gali labai stipriai ir greitai paveikti
kraujospūdį.
Neatlikite matavimo iš karto po fizinės veiklos ar išsimaudę. Prieš atlikdami matavimą pailsėkite nuo dvidešimties iki
trisdešimties minučių.
Nesukryžiuokite kojų. Padėkite pėdas ant grindų ir atsisėskite tiesiai.
Šis prietaisas matuoja pagal širdies ritmą. Jei jūsų širdies ritmas labai lėtas ar nereguliarus, gali būti sunku nustatyti
kraujospūdį prietaisu.
Jei prietaisas nustato neįprastą būklę, jis sustabdo matavimą ir parodo klaidos simbolį. Simbolių aprašymą rasite 6
puslapyje.
Šis kraujospūdžio matuoklis skirtas naudoti suaugusiesiems. Prieš matuodami kraujospūdį vaikui, pasitarkite su
gydytoju. Vaikas naudoti šio prietaiso be priežiūros negali.
Automatinio kraujospūdžio matuoklio veikimą gali paveikti per didelė temperatūra, drėgmė ar aukštis.
Lietuvių k.

8

4. Matavimas

Matuodami galite jausti didelį spaudimą rankovės srityje.

5. Po matavimo

Jei, kol ekrane pateikiami rodmenys, paspausite mygtuką START norėdami išjungti prietaisą, nauji rodmenys bus
išsaugoti atmintyje.
Jei, kol ekrane pateikiami rodmenys, paspausite mygtuką
norėdami išjungti prietaisą, nauji rodmenys bus
išsaugoti atmintyje.
Nuimkite rankovę ir įrašykite savo duomenis.
Pastabos:

prietaise esanti automatinio maitinimo išjungimo funkcija esamus duomenis išsaugo atmintyje ir
automatiškai išjungia prietaisą nuo matavimo praėjus maždaug vienai minutei. Prieš matuodami
kraujospūdį tam pačiam žmogui palaukite bent tris minutes.

Lietuvių k.

9Matavimai
Pastaba:vieną kartą panaudotas UB-525 pripūtimą pakoreguoja pagal naudotoją.

1. Apsukite rankovę aplink riešą. Patogiai atsisėskite taip, kad rankovė būtų širdies
lygyje, ir atsipalaiduokite.
2. Paspauskite mygtuką START . Ekrane rodomi visi segmentai.
3. Trumpai mirksi nulinė (0) reikšmė. Kai pradedamas matavimas, ekrano
rodmenys keičiasi. Rankovė pradeda pildytis oru. Normalu, kad matavimo metu
gali labai spausti rankovės sritį. Matavimas automatiškai prasideda prasidėjus
pūtimui, mirksi(širdies simbolis).
Pastaba:

jei nenorite, kad nauji rodmenys būtų išsaugoti atmintyje,
paspauskite mygtuką , kol matote rodmenis.

5. Dar kartą paspauskite mygtuką START , kad išjungtumėte prietaisą.
Nusiimkite rankovę.
Pastabos:

Nulinės reikšmės
ekranasekranasPrasidėjo pūtimas
pūtimasPrasidėjo

jei norite bet kuriuo metu sustabdyti pūtimą, dar kartą paspauskite
mygtuką START .

4. Kai matavimas baigtas, prietaise pateikiami rodmenys (sistolinio ir diastolinio
spaudimo rodmenys, pulso dažnis, PSO klasifikavimo ženklas, I.H.B. simbolis).
Rankovė išleidžia likusį orą ir visiškai subliūkšta automatiškai.
Pastaba:

Visi segmentai

Vyksta pūtimas ir
matavimas
Sistolinis spaudimas
Diastolinis spaudimas
PSO klasifikavimas
WHO
I.H.B. simbolis
Pulso dažnis

prietaisas turi automatinio maitinimo išjungimo funkciją.
Prieš matuodami kraujospūdį tam pačiam žmogui palaukite bent tris
minutes.

Lietuvių k.

10

Atminties duomenų atkūrimas
Pastaba:
šis prietaisas atmintyje gali išsaugoti paskutinius 60 matavimų.
1. Išjungdami prietaisą paspauskite mygtuką .
Parodomas visų matavimo rezultatų vidurkis ir duomenų numeris. Jei
duomenų nėra, rodomas 0. Norėdami išjungti prietaisą paspauskite
arba mygtuką START .
2. Norėdami pamatyti duomenis (duomenų numerį ir matavimo duomenis),spauskite nurodytus mygtukus.
Prietaisas parodo visų matavimo rezultatų vidurkį ir duomenų
numerį.
Kiekvieną kartą paspaudus mygtuką , prietaiso ekrane rodomi:
per laikotarpį atliktų matavimų duomenų vidurkis;
duomenys (numeris ir matavimo duomenys).
Prietaisas duomenis rodo nuo naujausiųjų.
Pavyzdys:„No.30“ ir duomenys„No.29“ ir duomenys„No.01“ ir duomenys.
paspaudžiamas parodžius seniausius duomenis,
3. Jei mygtukas
prietaisas pereina prie 1 veiksmo: rodomas visų matavimo rezultatų
vidurkis ir duomenų numeris.
4. Norėdami išjungti prietaisą, paspauskite mygtuką START .
Po vienos minutės neveikimo prietaisas išsijungs automatiškai.

Atminties režimas

Išjungti
MygtuskasVisų duomenų vidurkisVisų duomenų numeris
MygtukasPvz.: „No.30“
Naujausi duomenys
Mygtukas
Paskutiniai duomenys(seniausi)

Kaip pašalinti atmintyje saugomus duomenis
1. Paspauskite ir laikykite paspaudę mygtuką , kol prietaisas išsijungs
automatiškai.
, trinant atmintyje saugomus duomenis
Prietaiso ekrane pasirodys simbolis, tada prietaisas išsijungs automatiškai.
mirksės simbolis
Lietuvių k. 11

Mygtukas

Išjungti
Mygtukas

Kas yra IHB/AFib indikatorius?
Kai matuoklis matuodamas nustato nereguliarų ritmą, ekrane pasirodo IHB/AFib indikatorius su matavimovertėmis.
Pastaba:rekomenduojame kreiptis į gydytoją, jei dažnai matote šį

IHB/AFib indikatorių.

Kas yra AFib?

Širdis susitraukia dėl elektrinių signalų, atsirandančių širdyje, ir varinėja kraują po visą organizmą. Prieširdžių
virpėjimas (angl. AFib) atsiranda tada, kai elektrinis signalas prieširdyje sutrinka ir sukelia pulso dažnio sutrikimus.
AFib gali sukelti kraujo sąstovį širdyje, dėl kurio gali lengvai susidaryti kraujo krešulių, ištikti insultas ar širdies
priepuolis.
Vykdomas spaudimas

Spaudimo juostos indikatorius

Indikatorius stebi spaudimo eigą matuojant.
Vyksta pūtimas
Vyksta matavimas
Pūtimas baigtas

Lietuvių k.

12

PSO klasifikavimo indikatorius
Kiekvienas iš šešių juostos indikatoriaus segmentų atitinka
PSO kraujospūdžio klasifikaciją, apibūdintą 14 puslapyje.
Pavyzdys
Vidutinio sunkumo
hipertenzija

Nedidelė
hipertenzija

PSO klasifikavimo indikatorius
Sunki hipertenzija
Vidutinio sunkumo hipertenzija
Nedidelė hipertenzija
Viršutinė normos riba
Normalus
Optimalus

Viršutinė normosriba: indikatorius nurodo segmentą,remdamasis dabartiniais duomenimis
ir PSO klasifikacija.

Kas yra kraujospūdis?

Apie kraujospūdį

Kraujospūdis yra kraujo slėgis į arterijų sieneles. Sistolinis spaudimas susidaro širdžiai susitraukiant. Diastolinis
spaudimas susidaro širdžiai atsipalaiduojant. Kraujospūdis matuojamas gyvsidabrio stulpelio milimetrais (mmHg).
Natūralų žmogaus kraujospūdį atspindi pagrindinis slėgis, išmatuotas ryte tik atsibudus, ramybės metu ir priešvalgį.

Kas yra hipertenzija ir kaip ji kontroliuojama?

Hipertenzija, t. y. neįprastai aukštas arterinis kraujospūdis, jei nesekama, gali sukelti daugybę sveikatos sutrikimų,įskaitant insultą ir širdies priepuolį. Hipertenzija gali būti kontroliuojama keičiant gyvenimo būdą, vengiant streso ir
vartojant vaistus gydytojui prižiūrint.
Norint išvengti hipertenzijos arba ją kontroliuoti, reikia laikytis tolesnių nurodymų.
Nerūkyti.
Reguliariai mankštintis.
Sumažinti druskos ir riebalų kiekį maiste.
Reguliariai tikrintis sveikatą.
Išlaikyti tinkamą svorį.
Lietuvių k. 13Kodėl verta matuoti kraujospūdį namuose?

Kraujospūdžio matavimas klinikoje ar gydytojo kabinete pacientui gali sukelti nerimą ir gali būti gautas padidėjęs
rodmuo (25–30 mmHg didesnis, nei matuojant namuose). Kraujospūdžio matavimas namuose sumažina išorinių
veiksnių poveikį kraujospūdžio rodmenims, papildo gydytojo rodmenis ir užtikrina tikslesnį kraujospūdžio matavimą.

PSO kraujospūdžio klasifikacijaŽmogaus kraujospūdis kiekvieną dieną ir kiekvieną sezoną skiriasi.
Dėl įvairių sąlygų dienos metu jis gali kisti 30–50 mmHg.
Hipertenzija sergančių žmonių kraujospūdžio svyravimai dar
ryškesni. Paprastai kraujospūdis padidėja darbe ar žaidžiant,o miego metu jis sumažėja. Todėl nesikliaukite vieno matavimo
rezultatais.
Norėdami sužinoti savo įprastą kraujospūdį, matuokite kraujospūdį
kiekvieną dieną tuo pačiu metu, laikydamiesi šiame vadoveaprašytos tvarkos. Reguliariai matuojant kraujospūdį gaunama
išsamesnė kraujospūdžio istorija. Įrašydami kraujospūdį būtinaipasižymėkite matavimo datą ir laiką. Konsultuokitės su gydytoju dėlgautų kraujospūdžio duomenų interpretavimo.

Diastolinis kraujospūdis

Kraujospūdžio pokyčiai

3 laipsnio hipertenzija (sunki)
2 laipsnio hipertenzija (vidutinio sunkumo)
1 laipsnio hipertenzija (lengva)
Viršutinė normos riba
Normalus
Optimalus

Sistolinis kraujospūdis
Tipiški svyravimai per dieną
(matuojant kas penkias minutes)
Sistolinis

Kraujospūdis

Pasaulio sveikatos organizacija (PSO) nustatė standartus,
skirtus aukštam kraujospūdžiui įvertinti, neatsižvelgiant į amžių,kaip parodyta schemoje dešinėje.

Šaltinis: „Journal of Hypertension“,
1999 m., 17 tomas, Nr. 2

Diastolinis

PopietLietuvių k.

14

MiegasPriešpiet

LaikasTrikčių šalinimas
Problema

Galima priežastis
Maitinimo elementai išsikrovę.

Rekomenduojamas veiksmas
Pakeiskite visus maitinimo elementus naujais.
Iš naujo įdėkite maitinimo elementus taip, kad jų
neigiami ir teigiami poliai būtų nukreipti, kaip
pavaizduota maitinimo elementų skyrelyje.

Ekrane nieko
nerodoma net tada,
Maitinimo elementų poliai nukreipti
kai prietaisas įjungtas. netinkamai.
Rankovė nesipildooru.

Prietaisas nematuoja
kraujospūdžio.
Rodmenys yra per
dideli arba per maži.

Kita
Pastaba:

Maitinimo elementų įtampa per žema.
(simbolis MAITINIMO
MirksiELEMENTAI SENKA). Jei maitinimo
elementai visiškai išsekę, šis simbolis
nerodomas.
Rankovė uždėta netinkamai.

Pakeiskite visus maitinimo elementus naujais.

Matuojant pajudinote riešą ar kūną.
Rankovės padėtis netinkama.

Kraujospūdžio rodmenys skiriasi nuo
išmatuotų klinikoje ar gydytojo kabinete.

Taisyklingai uždėkite rankovę.
Sėdėkite tyliai ir ramiai, kol matuojamas
kraujospūdis.
Patogiai atsisėskite ir nusiraminkite.
Padėkite ranką ant stalo, pasukite delną į viršų ir
patikrinkite, ar rankovė yra širdies lygyje.
Jei jūsų širdies ritmas labai lėtas ar nereguliarus,
gali būti sunku nustatyti kraujospūdį prietaisu.
Žr. skirsnį „Kodėl verta matuoti kraujospūdį
namuose?“.
Išimkite maitinimo elementus. Vėl juos tinkamaiįdėkite ir bandykite matuoti dar kartą.

jei anksčiau aprašyti veiksmai neišsprendžia šių problemų, susisiekite su pardavėju. Nebandykite patys
atidaryti ar taisyti šio gaminio, nes garantija negalios.

Lietuvių k.

15

Techninė priežiūra
Neardykite prietaiso. Jame naudojami jautrūs elektriniai komponentai ir sudėtingas oro blokas, kurie gali būtisugadinti. Jei negalite išspręsti problemos, naudodamiesi trikčių šalinimo nurodymais, kreipkitės į įgaliotąjį
pardavėją savo teritorijoje arba į mūsų klientų aptarnavimo skyrių. „A&D“ klientų aptarnavimo skyrius suteiks
techninės informacijos, pateiks atsarginių dalių ir įrenginių įgaliotiesiems pardavėjams.
Prietaisas buvo suprojektuotas ir pagamintas taip, kad ilgą laikotarpį jam nereikėtų priežiūros. Paprastai
rekomenduojama prietaisą tikrinti kas 2 metus, kad būtų užtikrintas tinkamas veikimas ir tikslumas. Dėl techninės
priežiūros kreipkitės į įgaliotąjį pardavėją savo teritorijoje arba į „A&D“.

Techniniai duomenys
Tipas
Matavimo metodas
Matavimo diapazonas

Matavimo tikslumas
Maitinimo šaltinis
Matavimų skaičius
Riešo apimtis
Klasifikavimas
Darbinė dalis
Naudingo eksploatavimo trukmė

UB-525
Oscilometrinis matavimas
Spaudimas:
0 – 299 mmHg
Sistolinis spaudimas:
60 – 279 mmHg
Diastolinis spaudimas:
40 – 200 mmHg
Pulsas:40 – 180 dūžių per minutę
Spaudimas: ± 3 mmHg
Pulsas:±5%
Du 1,5 V šarminiai maitinimo elementai (LR03 arba AAA)Maždaug 250 matavimų, įdėjus AAA šarminius maitinimo elementus, kai
spaudimo vertė yra 170 mmHg, o patalpos temperatūra siekia 23 °C.
13,5–21,5 cm
Medicininė įranga su vidiniu maitinimo šaltiniu (nuolatinis veikimo režimas)
BF tipo rankovė
Prietaisas: 5 metai (naudojant šešis kartus per dieną)Lietuvių k.

16

Klinikinis tyrimas

Pagal ISO81060-2：2013
Klinikinio aprobavimo tyrimo metu K5 buvo naudotas 85 tiriamiesiems diastoliniam
kraujospūdžiui nustatyti.

EMD (Elektromagnetiniai
IEC 60601-1-2: 2014
trikdžiai)Atmintis
Paskutiniai 60 matavimų
Veikimo sąlygos
Nuo + 10 iki +40 °C / 15–85 % sant. drėgmė / 800–1060 hPa
Transportavimo / laikymo sąlygos Nuo - 20 iki +60 °C / 10–95 % sant. drėgmė / 700–1060 hPa
MatmenysMaždaug 56 (P) x 88 (A) x 21,5 (I) mm
SvorisMaždaug 90 g be maitinimo elementų
Patekimo į prietaisą apsauga
IP20
Pastaba:

tobulinant prietaisą specifikacijos gali būti keičiamos be išankstinio įspėjimo.
IP klasifikacija yra apsaugos laipsnis, kurį užtikrina korpusai pagal IEC 60529. Šis prietaisas yra
apsaugotas nuo kietų, 12 mm skersmens ar didesnių svetimkūnių, tokių kaip pirštas. Šis prietaisas
neapsaugotas nuo vandens.

Lietuvių k.

17

ATMINTINĖ

Lietuvių k.

18

A&D Company, Limited
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119
Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY
United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119
A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409
OOO A&D RUS
OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow,
Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66
A&D Technology Trading(Shanghai) Co. Ltd

爱安德技研贸易（上海）有限公司

中国 上海市浦东新区浦东大道138号永华大厦21楼A室 邮编200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599