Mobile human monitoring
Better Data • Better Decisions • Better Performance

The EQ02+ LifeMonitor
The EQ02+ is the new multiparameter body worn
sensor in the Equivital LifeMonitor range.
It measures clinical grade cardiorespiratory, temperature
and activity data from mobile people. Data is stored
on the sensor and can be transmitted wirelessly forviewing on a mobile phone or computer.
The EQ02+ is ideal for monitoring human physiology acrossa wide range of applications including sports and exercise
research, clinical trials, biofeedback, CBRN and militarytraining and deployment.

New Features•	2 shoulder strap belt design for high quality data during
extreme ambulation• Upgraded 32bit arm core processor•	Improved battery performance•	Improved body temperature measurement

Benefits• 	2 Leads of 256Hz ECG: Allows multichannel
processing and reduction in noise from theambulatory signal• 2 5Hz and 256Hz Tri Axis Accelerometer:
Configurable to the requirements of a specificapplication from motion and activity analysis
to detailed gait analysis• Wireless connection to external sensors includingcore temperature pill, skin temperature patch andgalvanic skin response sensor – All data automaticallytime synchronised and stored on the SEM•	
FDA and CE clearance: Peace of mind that the system
you are using meets the requirements of the rigorous
medical device standards

Measurements
ECG, HR,R-R interval

Accelerometer,
X,Y,Z, Activity

Respiratory
Rate

Body position/Movement

Skin
Temperature

Oxygensaturation*

Galvanic skinresponse*Bloodpressure*Core temperature
capsule/Dermal
Patch** Measuredusing acompatible ancillary sensor

www.equivital.co.uk

How it works
The LifeMonitor comprises 2 elements; the Sensor Belt and the Sensor
Electronics Module (SEM). The sensor belt features low profile sensors
embedded into a biocompatible fabric. The SEM docks into a cradleon the left hand side of the sensor belt where it processes, storesand transmits physiological data.

The Sensor Belt
The sensor belt is made from a breathable, lightweightfabric to ensure that it is comfortable for long-term use.
The two-shoulder strap unisex design provides a secureand comfortable fit for both male and female users.
Key Features• Unisex• Breathable, sports fabric• Available in Sizes 29in – 47in• Washable

The SEMThe SEM slots into the low profile belt cradle andit’s LEDs, visible when worn on body, provide status
information including transmission, power andexternal sensor connections. It has a rechargeable
Li Polymer battery which takes approximately an hour
to reach full capacity. The SEM includes a medical grade
infra-red thermometer and tri-axis accelerometer.
All data is stored on an integral 8GB SD card,allowing retrospective data download and analysis.

Real time data transmission
Data can be transmitted in real time from the SEM, via Class 1 Bluetooth, to a mobile phone or computer, for viewing
and analysis.
Training and Research

SEM

BlackGhost System

BluetoothSEM
EqView Mobile

EqView Professional

Bluetooth/3G/4G
EqView Mobile

BlackGhost serverRadio Secure
Communications

BlackGhost serverBluetooth

SEM

USB
BluetoothBluetooth Dongle

EqView
Professional

www.equivital.co.uk

Software
Equivital Manager
Equivital Manager is a software application forviewing SEM status information, downloading andsplicing SEM files, pairing Bluetooth ancillaries andextracting data to CSV or Excel files. It is particularly
useful for managing data and settings in a small
number of SEMs, over USB.

Equivital Qiosk
Equivital Qiosk is a software application that is intended for use
in clinical trials and Black Ghost deployments where more SEMs
and data need to be managed. It has a pre-configured set up for
configuring and downloading data from a SEM, which is initiatedwithout the need for user interaction. This is especially suitable
where you need to maintain a consistent operating protocol
between multiple researchers, sites or subjects.

eqView Mobile
eqView Mobile is an Android app that allows data from a SEM to beviewed in real time. It requires a one to one connection over Bluetooth.
Key data including ECG can be viewed on the app and transmitted
on to either eqView Pro or Black Ghost.

www.equivital.co.uk

eqView ProeqView Pro is a software tool for viewing live data including
ECG and respiratory waveforms. Subjects can be assigned
sensors, multiple sensors can be time synchronised andsession event markers can be created during the session.

Black Ghost
Black Ghost is a web application tailored to the needs of CBRN, military and emergency first response users. It combines
SEM and geolocation data enabling small and large teams to be monitored in training or deployment missions.

System Developers Kit (SDK)Supported by our excellent in house team, Equivital offers developers a choice of .NET and Java SDK to engineer innovative
solutions with the EQ02+ LifeMonitor.
www.equivital.co.uk

EQ02+ LifeMonitor key specifications
Parameter

Specification

Physical specifications

SEM

Memory8GB

Weight38g

Power

Li-Po rechargeable cellSize

78mm x 55mm x 11mm

Li Polymer
recharge timeRecharge from flatin 1 hour

Operating temperature-10 to +50oCOperating humidity

Communications

Class 1 Bluetooth 2.1, 100m
operating range

0 to 95% RH
Non-condensing

Operating altitude-300ft to 30,000ft

Connectivity

USB 2.0 compatible

Water ingress protection

IPx7

Configuration

Via USB

Signals

Sampling rate

Resolution

ECG

256Hz

10 bitRespiratory

25.6Hz

10 bitSkin temperature

15 seconds

0.1oC

Accelerometer

25.6 or 256Hz

12 bitPhysiologicalparameters

Reporting range

Heart rate

25-240bpm

Breathing rate
Skin temperature

Body position

Approvals
EuropeCE Certified, EU Device
Classification Class IIb

0-70 Breaths per minuteUSA-10oC to +50oC±0.3oC Accuracy using
IR Thermometer attemperature of interest

FDA Device Classification
Class II

Health Canada

Device Class II

All orientations
Includes fall detection
and motion

Battery Life Data
SEM use caseOperating life

Operating life with external
battery pack connected

All SEM data recorded

48hrs

138hrsAll SEM data recordedwith core temperature pill receiver on

18hrs

51hrs

All SEM data recorded
and transmission over bluetooth on

12hrs

38hrs

For further information or support on Equivital products please contact the Equivital team at info@equivital.co.uk

www.equivital.co.uk

Equivital Ltd
Unit F, Buckingway Business Park, Swavesey, Cambridge CB24 4UQ, UK
T: +44 (0) 1954 233430 F: +44 (0) 1954 233431 E: info@equivital.co.uk W: www.equivital.co.uk
HIDA3330- DS-EQ02PLUS-1p0

www.equivital.co.uk