Deep UV Lasers224.3 and 248.6nm

Features
Deep Ultraviolet For Less
The DUV family of lasers offer 224.3 nm and 248.6 nm for fraction ofthe cost of the competition. The laser is the size, weight and power
consumption of a HeNe laser but with output in the deep UV. The selfcontained, integrated, laser controller enables remote computer
control for ease of operation and flexible data collection via LabView
software. With an input power less than 10 W the need for watercooling and other thermal management issues is eliminated. The lasers
reach full power in less than 20 microseconds from a cold start fromany ambient temperature from -10 to 40 °C without preheating ortemperature regulation. With output over 100mW and linewidths lessthan 3GHz or 0.0005nm these are great sources for a wide range ofapplications.
Ultra-easy ultraviolet. Make ultra-sensitive measurements of Raleigh,Raman, fluorescence or phosphorescence emissions generated by deep
UV excitation. Our “instrument solutions” combine a deep UV laser
source with an array of analyzer and detector plug-and-play modules.
Detection choices include single and multi-channel PMT andphotodiode detector modules that are gated in synchronism with thelaser and offer flexible boxcar integration and averaging for enhanced
signal-to-noise data collection. Analyzer options include UV Raman,laser induced native fluorescence (autofluorescence), CE and HPLC
modules. The combination of plug-and-play source, analyzer anddetector modules enables you to rapidly develop breadboard andprototype instruments for a wide range of applications from research
to product analysis to environmental monitoring with data sampling
rates up to 20 Hz.
Flexible for the lab, made for the real world. An array of accessories
such as emission line purity modules and fiberoptic couplers enable
you to mate our components with a wide range of devices from thirdparty suppliers. Communication with the laser and all plug-and-playmodules is accomplished via USB or Ethernet using LabView drivers.
Our "instrument solutions" provide a seamless fit for many applications
such as laser induced native fluorescence or UV resonance Raman
analyzers, photoluminescence, capillary electrophoresis, highperformance liquid chromatography, phosphorescence and many othertypes of instruments. Rugged design, reliable performance and low
cost make them ideal for the field researcher and the OEM.

1512 Industrial Park St., Covina, CA 91722

T: 626 967-6431

F: 626 967-5813• Wavelengths from 224 – 248 nm,> 200 mW
• Innovative plug-and-play instrument
solutions• Rugged design for reliability• Wide operating environment (-10 to40 °C)• Square pulse width from 20 – 120 μs• “Soft” pulse output reduces thermaldamage• Narrow line width < 3 GHz (0.0005nm, 0.5 pm, 0.1 cm-1)• Instant on (< 10 μs from cold start)• 90VAC–240VAC input<10W average,
no water cooling or toxic chemicals• USB or Ethernet interface withLabView• Built-in laser power monitor• CE and RoHS
www.photonsystems.com

HeAg lasers @ 224.3nm
Model
Peak power (quasi-cw)
System dimensions
System weight
Pulse frequency
Longitudinal mode spacing
Pulse width
Pulse synchronism
Beam diameter
Beam divergence
Oscillation bandwidthPower consumption
Line requirements

HeAg 70-224SL
HeAg 30-224SL>50 mW>10 mW
10x10x70 cm
5x13x30 cm
3.6 kg
1.4 kg1 – 20 Hz
1 – 5 Hz257 MHz
642 MHz
20 μS to 120 μS, adjustableinternal or external
3 mm< 4 mrad< 3 GHz, < 0.10 cm-1
< 10 W
90-250VAC, 47-63 Hz, 100mA or 24VDC@400mA

Series 30 Laser with integratedcontroller

NeCu lasers @ 248.6nm
Model
Peak power (quasi-cw)
System dimensions
System weight
Pulse frequency
Longitudinal mode spacing
Pulse width
Pulse synchronism
Beam diameter
Beam divergence
Oscillation bandwidthPower consumption
Line requirements

NeCu 70-248
NeCu 30-248>250 mW>50 mW
10x10x70 cm
5x13x30 cm
3.6 kg
1.4 kg1 – 20 Hz1 – 5 Hz
257 MHz642 MHz
20 μS to 50 μS, adjustableinternal or external
3 mm< 4 mrad< 3 GHz, < 0.10 cm-1
< 10 W
90-250VAC, 47-63 Hz, 100mA or 24VDC@400mA

1512 Industrial Park St., Covina, CA 91722

T: 626 967-6431

F: 626 967-5813

Series 70 Laser with integrated
controller

www.photonsystems.com