Proceq® Flaw Detector Live

UT8000

Flaw detection of metal and composite components using
ultrasonic testing

Versatility

Collaboration

Peace of Mind

Ultralight, heavy-duty, IP67.
Flexible screen size forunmatched portability. Your
companion of choice for themost demanding inspection
jobs.

Intelligent software lets you
annotate measurements withvoice, photos, and comments.
Generate reports and sharethem instantly. Access your datafrom anywhere, anytime.

The only flaw detector that canrewind time to replay theinspection process. Automatedlogging of measurement settings
finally makes your procedures
traceable.

Proceq® UT8000
GeneralCalibration
Automated calibration

Velocity, zero offset

Test modes

Pulse echo, dual, through transmission orgridUnits

Millimeters, inches, or microsecondsRange

3 – 11760 mm at 5900 m/s

LEMO 00

Velocity

100 – 10000 m/s

Zero offset

Data storage

Up to 1 terabyte (TB), depending on iPadmodel

0 – 50 mm

Display delay

up to full range

Refracted angle

0° – 90° in 1° increments

Battery type and life

6x AA rechargeable batteries; 5 hautonomy; compatible with off-the-shelf
USB power banks

Power requirements

Universal AC (100-240 V, 50-60 Hz)

Gates
Measurement gates

2x, fully independent

Display type2048 × 1536 / 326 PPI; fully laminated with
antireflective coating and up to 120 Hz
refresh rate

Gate start

Variable over entire displayed range

Display dimensions

7.9" – 12.9" diagonal

Gate width

Variable, from 0.30 Ɋs to end of displayed
range

Overall Dimensions (W × H × D)

215 mm × 101 mm × 36 mm
8.46” x 3.97” x 1.42”

Gate height

Variable, 1% – 100% full screen height, inincrements of 1%

Weight698 g, incl. battery pack1.54 lbs incl. battery pack

AlarmsPositive and negative threshold/curve

USB ports

2x USB-C

Video output

Digital; wireless over Apple TV;
HDMI/VGA/DisplayPort (via third-partyadapter)

Proceq® UT8000 (793 10 500)

Base unit, battery pack, batteries, power
supply, belt holder, screwdriver,documentation, carrying caseCompliance

EN 12668-1:2010; ASTM E317-16

User Interface Languages

English, Spanish, French, German,
Japanese, Chinese, Korean

Transducer connections

Environmental ratings
IP rating

IP67

Shock tested

MIL-STD-810F; Method 516.5; Procedure I,
6 cycles each axis; 15 g; 11 ms half sine

Vibration tested

MIL-STD-810F; Method 514.5; Procedure I,Annex C; Figure 6; general exposure; 1 h /axis

Operating temperature Proceq UT8000 -10 to 50 °C (14 to 122 °F)
Battery operating temperature

0 to 40 °C (32 to 122 °F)

Battery storage temperature

0 to 50 °C (32 to 122 °F)

Measurements
Measurement display locations

Up to 5 locations (manual or auto selection)

Gate (1, 2)

Thickness, sound path, projection, depth,
amplitude, time-of-flight, min./max. depth,min./max. amplitude, sizing measurements
based on mode

Echo-to-echoStandard gate 2 – gate 1

DAC/TCG

Standard, up to 10 points, 110 dB dynamic
TCG range

Special

DAC modes: Custom DAC (up to 6 curves),
20% – 80% view

Software sourcing options

PulserPulserTunable square wave

Pro (793 65 150)

Initial subscription (24 months)

PRF

10 – 2000 Hz, in increments of 10 Hz

Pro renewal (793 65 151)

Subsequent yearly subscription fee

Energy settings

50, 100 or 400 V

Pulse width

Adjustable; 25 – 2500 nsec

Damping

50, 400 ȍ

Accessories
Spare battery pack (793 30 110)

ReceiverGain

0 – 110 dB

Receiver input impedance

400±5% ȍReceiver bandwidth

0.25 – 20 MHz at -3 dB
EN12668 compliant

Digital filter settings

8 digital filter sets

Rectification

Full-Wave; Positive Half-Wave; Negative
Half-Wave; RFSystem linearity

Horizontal, ±0.5% FSW

Sampling frequency

125 MHz

Reject0 – 100% FSH, in increments of 1%

Amplitude measurement

0.1% – 125% full screen height

Measurement rate

Equivalent to PRF in all modes (single shot)

Tablet holder (793 30 120)
Supporting accessories

Battery pack including 6xAA rechargeablebatteries
iPad Mini – iPad Air
Wide range of ultrasonic transducers, cables
and calibration blocks, please call your local
representative for more information.

iPad is a trademark of Apple Inc.

www.proceq.com