AAFM Approved Apparatus must be installed in accordance with manufacturer’s instructions.
Use supply wires suitable for 5°C above surroundings.
Only simple apparatus should be terminated to the sensor connection.
Simple apparatus are components as defined by the NEC (1.2 V, 0.1 A, 0.25 mW or 20 µJ).
Warning: Substitution of components may impair intrinsic safety or suitability for Class I, Division 2.

EXPLOSION PROOF
XPClass I / Div. 1 / Groups ABCD
DUST IGNITION PROOF
DIP Class II, III / Div. 1 / Groups EFG- Install per National Electrical Code (NFPA 70)- For Group A, seal all conduits within 18 inches of enclosure; otherwise, conduit seal not required for compliancewith NEC 501.5(A)(1)(1).
- All conduits must be assembled with a minimum of five full threads engagement.
- Temperature sensor assembly must be FM approved for appropriate area classification.
- Class II use a dust tight seal
- Keep tight when circuits alive

e.g. Remote mount sensor configuration+ power supply
Nonhazardous Locations

Hazardous (Classified) Location
Class I / Division 2 / Groups ABCD

5Installation Notes TMT84, TMT85-FM explosionproof approved
temperature sensor assembly

4321

Nonhazardous Locations

Hazardous (Classified) Location
Class I / Division 1, 2 / Groups ABCDClass I / Zone 1 / IIC T6/T5/T4
Class II / Division 1, 2 / Groups EFG
Class IIIB

B+-e.g. RTD or TC Sensor(Simple Apparatus)
integral or remote mounted

FM Approved Associated Apparatus
or Associated Nonincendive Field
Wiring Apparatus

Temperature range

C

without display, TID10
T4-40°C ... +85°C
T5-40°C ... +80°C
T6-40°C ... +70°C

with display, TID10T4-40°C ... +85°C
T5-40°C ... +80°C
T6-40°C ... +70°C

NONINCENDIVE, FIELD WIRINGNI Class I / Div. 2 / Groups ABCD

Sensor circuits (Terminals 3…7)

NOTE
When the product is installed as a FNICO installation use drawing 34 02 00 111.

Uo or Voc or Vt = 7.2 V

Io or Isc = 25.9 mA

Po = 46.7 mWGroup A, B resp. IIC
Group C, D resp. IIB
Group C, D resp. IIA

Co or Ca = 13.5 µF
Co or Ca = 240 µF
Co or Ca = 1000 µF

Lo or La = 59 mH
Lo or La = 238 mH
Lo or La = 477 mH

Pfanzelt
Volume (mm³)

DesignedPfanzelt
Refer to protection notice
ISO 16016

Drawing No.

Date (yyyy-mm-dd)

2011-06-08
Date (yyyy-mm-dd)

34 02 00 113

Unit

2011-06-06

Edge of working parts Geometrical tolerancing
ISO 2768-mH-E
ISO 13715

Dwg.rev.

-ScaleiTEMP TMT84, TMT85
Part No.

-

1:1
Format

A4

Revision no.

-71540295

Revision date (yyyy-mm-dd)

Name

Material--

XA02281T/09/EN/01.20

Title

CONTROL DRAWING FM
XP, NI, DIP

D

DApproved

C

NONINCENDIVEClass I / Div. 2 / Groups ABCD
AEx nA II- Intrinsic safety barrier not required. Vmax £ 35 V DC.
- Warning: Do not disconnect equipment unless power has been switched off or the area is known to be nonhazardous.
- Nonincendive field wiring installation
The Nonincendive Field Wiring Circuit Concept allows interconnection of Nonincendive Field Wiring Apparatus
with Associated Nonincendive Field Wiring Apparatus or Associated Intrinsically Safe Apparatus or AssociatedApparatus not specifically examined in combination as a system using any of the wiring methods permitted forunclassified locations, whenVoc £ Vmax, Ca ³ Ci + Ccable, La ³ Li + Lcable.
Transmitter Nonincendive Field Wiring parameters are as follows: Ui or Vmax £ 35 V DC Ci £ 5 nF Li £ 10 mF
For these current controlled circuits, the parameter Imax is not required and need not to be aligned withparameter Isc and It of the Associated Nonincendive Field Wiring Apparatus or Associated Apparatus.
- Warning: Explosion Hazard- Do not disconnect equipment unless power has been switched off or the area isknown to be non-hazardous- The transmitter is suitable to be installed according the FNICO concept.

SeriesObjekt version

Sheet

Endress + Hause r
WetzerNesselwang / Germany

1 of 1 GmbH+Co. KG

54321