X1
https://www.gigahertz-optik.com/en-us/product/x1/

Product tags: Multi-Channel , Handheld device

Gigahertz Optik GmbH

1/17

Description
Hand-held Meter
The X1 optometer is one of the most versatile hand-held lightmeasurement instruments available. It combines a powerful electronic
design packaged in a light-weight ergonomic housing. Its compact size
makes it ideal for field service applications. A unique feature of the X1 is its
capability to operate detector heads housing up to four photodiodes withall four signals displayed on the four line display with on/off backlighting.

Simple to Use

X1₁ Hand-held meter

Operating the X1 is simple. The meter set-up is supported by an easy touse menu. The menu allows selection of the operating mode, the detector
and measurement parameter. Once set-up all settings are stored andrecalled on next power-up unless reinitialized. Measurement values are
displayed in absolute quantities for the particular detector connected.

X1 detector compatibility
The X1-1 optometer is compatible with all Gigahertz Optik detectors fittedwith a type -4 connector. This includes devices incorporating up to 4
photodiodes. The calibration data for detectors with type -4 connectors is
programmed into the X1-1 optometer. Therefore, X1-1 optometers arematched with specific detectors.
The X1-5 optometer is compatible with all Gigahertz Optik detectors fittedwith a type -5 connector. Only single-photodiode based detectors can besupplied with a type -5 connector. The calibration data is stored in an
EEPROM within the -5 connector. The X1-5 optometer reads thiscalibration data automatically. All detectors with -5 connectors are fullyinterchangeable with any X1-5 meter.
The X1-6 optometer is for use with the Gigahertz Optik MDC4-xxx range ofsmart detectors only.

Battery or USB powered
For on-site applications the X1 is operated with two standard 1.5 V AA
batteries. In remote control operation the X1 is powered through the USB
interface.

Four-channel Meter
The unique feature of the X1-1 is the capability to operate multi-cell
detector heads with up to four photodiodes with all four signals displayedor read-out via the USB interface.

Gigahertz Optik GmbH

2/17

Multipurpose Light Measurement Instrument
The X1 can be combined with most of the Gigahertz-Optik single cell or
multi cell light detector heads for use in a wide application range ofradiometric, photometric and colorimetric measurements

Interfaces
The X1 features a USB interfaces.

Specifications
GeneralShort description

Optometer for the individual configuration as photometer, RGB-luminous Color meter, Radiometer, UVradiometer, LASER-powermeter etc. with the detector heads supplied.

Main features

Compact gauge in ergonomic design for one Hand control. Four measurement chanels in multiplex
operation for use with single and multi-chanel detectors heads. Back illuminated Display with four lines.
Battery powered with two AA cells. USB-Interface. User Software and Software development
kit available.

Measurement range

Wide dynamic current measurement range from 0.1 pA (noise equivalent signal) up to 200 µA. Seven
gain ranges with manual or automatic selection. Measurement range in absolute units with given by thedetector heads responivity and calibration.

Typical applications

Measurement device for mobile use: measuremen of the illuminance condition, measurement of lamp
aging, etc. Because of ist USB Interface and the Software development kit the device can be integrated inremote applications.

Calibration

Calibration and comparison of the current responsivity in each of the seven gain ranges. Memory
for calibration data of several detector heads for measuremen in the absolute unit of the selected
detector head.

Product
Detector interface

9pin MDSM9 socket, 4 inputs

Measurement range

Seven (200 μA to 0.1 pA) manual or auto range which can be set by the user. The default setting
depends on the customer specific configuration.
Range Nr.

Gigahertz Optik GmbH

Range

Slew-Rate

Resolution ± calibration
uncertainty *)

T
P

max.

(10 - 90)%0

200.0 µA ***)

3 ms

±0.1 µA ± 0.3%<
2

1

20.00 µA

3 ms

±0.01 µA ± 0.3%<
2

(at 24 °C)

3/17

2

2.000 µA

3 ms

±0.001 µA ± 0.3%<
23

200.0 nA

3 ms

±0.1 nA ± 0.3%<
1

4

20.00 nA

3 ms

±0.01 nA ± 0.3%<152.000 nA

30 ms

±0.001 nA ± 0.4%<
16

200.0 pA

30 ms

±0.1 pA ± 0.4%<
1±Bias current (max.1,0pA) ***) The measurement uncertainty is usually below the calibration uncertainty, but must also be
considered. At very low currents, a detailed examination of the measurement uncertainty analysis is
recommended, because in this case the measurement uncertainty can predominate.
**) internal zero adjustment can reduce the bias current. Maximum zero adjusted bias current = ±0,2pA.
***) only for instrument versions with 7 measuring ranges****) valid for currents above 10pA
CW integration time

1 ms – 1 s

Sampling rateinternal sampling rate ADC 250 µs

Offset correction

Correction range transcending

Parameter adjustment

Remote control or front panel buttons (menu), set values permanently stored (EEPROM)

Calibration

X1-1: max. 256 data sets (one data set can include up to 4 channel entries) total stored in device
EEPROM
X1-5: max. 64 entries for one channel stored in detector EEPROM

Gigahertz Optik GmbH

4/17

Menu guide

Menu item
1. Mode2. Setup

3. Detector
4. Offset
5. Range

Dose

Submenu item
CW

Function
Measures respective of any offset
and calibration factors
programmed
Dose
Accumulates the single readings
as exposure for measured
quantity
CIE Yxy & T
Measures the CIE Color Values Yxyand T
CIE Yuv & T
Measures the CIE Color Values Yuv
and T
Zero Adjust
Performs a zero adjustment of theinternal amplifier and ADC
Integration
Sets the measurement(integration) time
Dose Time
Sets max. dose measurement time
Selects calibration data to calculate the measurement result
Performs an automatic offset adjustment("Offset = CW" or "Offset = 0")
Sets the measurement range(auto, manual)

The maximum dose to be displayed on the X1-5/1 device is 99999 TJ/cm² (for W/cm²) or 99999
TJ/m² (for W/m²)
Maximum time to be set for dose measurement: 255h 59min 59s

Miscellaneous
Power Supply

Two AA batteries ~ 250 hrs. operation time - backlit display off
Powered by USB-Interface

Interface

USB V1.1 (HID Device)

Temperature range

Operating: (5 to 40) °C
Storage: (-10 to 50) ° C

Humidity<80%, non-condensing

Display

LCD graphic display 97 x 32 pixel
Display area 14.3 mm x 35.8 mm
Switchable LED-backlight
Text display 4 rows each 14 characters

Front panel control

3 buttons, menu system

Dimensions

145 mm x 63 mm x 30 mm
compatibel

Weight

150 g

Info

Regular recalibration of the current calibration is recommended. Especially when very small
measurement currents have to be measured. In the case of very high humidity, fault currents of theradiometer are possible at low measuring currents and should be taken into account.

Configurable with

Gigahertz Optik GmbH

5/17

Product Name

Description

Go to productLP-9901

Detector head to measure Laser radiant power in W and Laser
irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/lp-9901/

VL-3701

Detector head for the measurement of photopic illuminance in
Lux [lx]

https://www.gigahertzoptik.com/enus/product/vl-3701/

VL-3702

Detector head for the measurement of photopic illuminance in
Lux [lx]. Class B, f1 ≤ 6 %https://www.gigahertzoptik.com/enus/product/vl-3702/

VL-3704

Detector head for the measurement of photopic illuminance in
Lux [lx]

https://www.gigahertzoptik.com/enus/product/vl-3704/

VL-3705

Detector head for the measurement of scotopic illuminance in
Lux [lx]

https://www.gigahertzoptik.com/enus/product/vl-3705/

PD-9310A

High sensitive detector head for the measurement of photopic
illuminance in Lux [lx]. Features: f1 ≤ 3 %, 2.8nA/lx, 20mm
diffuser, for the usage with optometers and amplifiers,
calibration certificate

https://www.gigahertzoptik.com/enus/product/pd-9310a/

PD-9310B

High sensitive detector head for the measurement of photopic
illuminance in Lux [lx]. Features: f1 ≤ 6 %, 2.8nA/lx, 20mm
diffuser, for the usage with optometers and amplifiers,
calibration

https://www.gigahertzoptik.com/enus/product/pd-9310b/

PD-9310B-N

Very high sensitive detector head for the measurement ofphotopic illuminance in Lux [lx]. Features: f1 ≤ 3 %, 28nA/lx, nodiffuser, for the usage with optometers and amplifiers,
calibration

https://www.gigahertzoptik.com/en-us/prod
uct/pd-9310b-n/

VL-3701 with SRTM37-L

Detector head to measure the photopic illuminance in lx and theluminance in cd/m²

https://www.gigahertzoptik.com/enus/product/vl-3701
with srt-m37-l/

PD-9310 with SRTM37-L

High sensitive detector head to measure the photopic luminancein cd/m². Features: front lense for 1°, 2°, 5° or 10° viewing angle,
for the usage with Optometers and amplifiers, calibration
certificate

https://www.gigahertzoptik.com/enus/product/pd-9310
with srt-m37-l/

Gigahertz Optik GmbH

Product Image

6/17

Product Name

Description

Go to productCT-4501

RGB detector head for photopic illuminance and luminous color.
Features: four sensor design, xy, u’v’, CCT, for the usage with X1and P-9801 Optometer, part of HCT-99D, calibration certificate,for polychromatic light,

https://www.gigahertzoptik.com/enus/product/ct-4501/

LDM-9810

Detector head to measure the photopic spot luminance in cd/m².
Features: selectable 20', 1° and 6° viewing angles, view finder,
focus able achromatic lens, for the usage with Optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/ldm-9810/

VL-1101

Photometric detector head with VL-11 mount. Features: modular
detector for use with integrating spheres, front lenses etc. For
use with optometers and signal amplifiers

https://www.gigahertzoptik.com/enus/product/vl-1101/

LDM-9901

Detector head to measure the photopic spot luminance in cd/m²

https://www.gigahertzoptik.com/enus/product/ldm-9901/

S-SDK-X20

Software Development Kit for X20 variants (X1 and HCT99).

https://www.gigahertzoptik.com/enus/product/s-sdk-x20/

S-X1

Application software for X1 variants.

https://www.gigahertzoptik.com/enus/product/s-x1/

VL-1101 +UMPA-0.5-11-RD
Detector head

Module detector head for the measurement of photopic
illuminance in Lux [lx]. Features: UMPA adapter for usage withintegrating spheres, for the usage with optometers andamplifiers, calibration certificate

https://www.gigahertzoptik.com/en-us/prod
uct/vl-1101uumpa-0511-rd/

ISD-5-VL

Integrating sphere detector for luminous flux (lm) of 2π spot
sources

https://www.gigahertzoptik.com/enus/product/isd-5-vl/

ISD-10-VL

Integrating sphere detector for luminous flux (lm) of 2π spot
sources

https://www.gigahertzoptik.com/enus/product/isd-10-vl/

Gigahertz Optik GmbH

Product Image

7/17

Product Name

Description

Go to product

ISD-15P-VL

Integrating sphere detector for luminous flux (lm) of 2π sources

https://www.gigahertzoptik.com/enus/product/isd-15p-vl/

TD-11VL01

Photometric, temperature stabilized detector with DP-11 mounthttps://www.gigahertzoptik.com/enus/product/td-11vl01/

RW-3701

Detector head for the measurement of irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/rw-3701/

RW-3702

Detector head for the measurement of irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/rw-3702/

RW-3703

Detector head for the measurement of irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/rw-3703/

RW-3704

Detector head for the measurement of irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/rw-3704/

RW-3705

Detector head for the measurement of irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/rw-3705/

RW-3708

Detector head for the measurement of irradiance in W/m²

https://www.gigahertzoptik.com/enus/product/rw-3708/

UV-3701

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 315-400nm (UV-A),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3701/

UV-3702

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 280-315nm (UV-B),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3702/

UV-3703

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 200/250-280nm(UV-C), cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3703/

Gigahertz Optik GmbH

Product Image

8/17

Product Name

Description

Go to productUV-3710Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 320-400nm (UV-A),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3710/

UV-3711

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 280-320nm (UV-B),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3711/

UV-3716

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 305-400nm (UV-A),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3716/

UV-3717

Detector head for the measurement of irradiance of UV radiationin W/m²

https://www.gigahertzoptik.com/enus/product/uv-3717/

UV-3719

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 250-400nm (UV),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3719/

UV-3720

Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 240-320nm (UV),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3720/

UV-3721Detector head for the measurement of irradiance of UV radiationin W/m². Features: spectral responsivity from 350-400nm (UV-A),
cosine field-of-view, for the usage with optometers andamplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3721/

UV-3718

Detector head for the measurement of high irradiance of UV-C
254nm radiation in W/m²

https://www.gigahertzoptik.com/enus/product/uv-3718/

ISD-5-VISNIR

Integrating sphere detector for radiant power in W of 2π sources

https://www.gigahertzoptik.com/enus/product/isd-5-visnir/

ISD-3P-Si

Integrating sphere detector for Laser power in W

https://www.gigahertzoptik.com/enus/product/isd-3p-si/

Gigahertz Optik GmbH

Product Image

9/17

Product Name

Description

Go to productUV-3706

Detector head to measure irradiance W/m² in Bilirubin
phototherapy. Features: Bilirubin actinic responsivity, cosine fieldof-view, for usewith optometers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/uv-3706/

UV-3711-308

Detector head for the measurement of irradiance of 308nm
Eximer Lasers in W/m². Features: flat spectral responsivity besite
308nm. cosine field-of-view, dose measurement in conjuntion
with P-9710 optometer, calibration certificate

https://www.gigahertzoptik.com/enus/product/uv-3711-2/

UV-3709

Detector for Blue-light hazard measurements. Features: Singlecell detector, BLH actinic irradiance, for the use with optometer,
calibration certificate

https://www.gigahertzoptik.com/enus/product/uv-3709/

UV-3725

Detector for the measurement of UV-C 254 nm irradiance in air
disinfection applications

https://www.gigahertzoptik.com/enus/product/uv-3725/

ISD-3P-IGA

Integrating sphere detector with InGaAs photodiode and 30 mm
sphere for Laser power in W.

https://www.gigahertzoptik.com/en-us/prod
uct/isd-3p-iga-2/

ISD-5-Si

Integrating sphere detector for Laser power in W

https://www.gigahertzoptik.com/enus/product/isd-5-si/

RCH-102

Detector head for high intensity irradiation in UVA and blue lightcuring processes with rigid fiber

https://www.gigahertzoptik.com/enus/product/rch-1/

RCH-116

Detector head with rigid fiber for high intensity UV and BLUE LED
sources.

https://www.gigahertzoptik.com/enus/product/rch-2/

ISD-5P-Si

Integrating sphere detector for Laser power in W

https://www.gigahertzoptik.com/enus/product/isd-5p-si/

ISD-10-Si

Integrating sphere detector for Laser power in W

https://www.gigahertzoptik.com/enus/product/isd-10-si/

ISD-15-Si

Integrating sphere detector for Laser power in W

https://www.gigahertzoptik.com/enus/product/isd-15-si/

Gigahertz Optik GmbH

Product Image

10/17

Product Name

Description

Go to product

RCH-006

Detector head for high intensity irradiation in UV wide range
curing processes

https://www.gigahertzoptik.com/enus/product/rch-006/

RCH-008

Detector Head for High-Intensity Irradiation in UV-A Curing
Processes

https://www.gigahertzoptik.com/enus/product/rch-008/

RCH-009

Detector Head for High-Intensity Irradiation in Blue Light Curing
Processes

https://www.gigahertzoptik.com/enus/product/rch-3/

RCH-010Detector head for high intensity irradiation in UV H-type lightcuring processes.

https://www.gigahertzoptik.com/enus/product/rch-4/

RCH-011

Detector head for high intensity irradiation in UVA peak lightcuring processes.

https://www.gigahertzoptik.com/enus/product/rch-5/

RCH-012

Detector head for high intensity irradiation in blue light curing
processes.

https://www.gigahertzoptik.com/enus/product/rch-6/

RCH-013

Irradiance Detector for UV or Blue light curing processes

https://www.gigahertzoptik.com/enus/product/rch-7/

RCH-014

Detector head for high intensity irradiation in UV or blue lightcuring processes. Features: Separate light integrator anddetector with flexible fiber coupling, 400nm+436nm BLUE
responsivity, wide viewing angle, for the usage with optometers
and amplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/rch-8/

RCH-015

Detector head for high intensity irradiation in UV or blue lightcuring processes. Features: Separate light integrator anddetector with flexible fiber coupling, light, 436nm BLUE-Peakresponsivity, wide viewing angle, for the usage with optometers
and amplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/rch-9/

RCH-106

Detector head for high intensity irradiation in UV wide range
curing processes

https://www.gigahertzoptik.com/enus/product/rch-10/

PD-11 Series

Detector head with DP-11 mounthttps://www.gigahertzoptik.com/en-us/prod
uct/pd-11-serie/

Gigahertz Optik GmbH

Product Image

11/17

Product Name

Description

Go to productRCH-108

Detector head for high intensity irradiation in UVA Peak lightcuring processes

https://www.gigahertzoptik.com/enus/product/rch-11/

RCH-109

Detector head for high intensity irradiation in blue-peak lightcuring processes

https://www.gigahertzoptik.com/enus/product/rch-12/

RCH-110Detector head for high intensity irradiation in H-Type light curing
processes

https://www.gigahertzoptik.com/enus/product/rch-13/

RCH-111

Detector head for high intensity irradiation in UVA light curing
processes

https://www.gigahertzoptik.com/enus/product/rch-14/

RCH-112

Detector head for high intensity irradiation blue light curing
processes.

https://www.gigahertzoptik.com/enus/product/rch-15/

RCH-113

Detector head for high intensity irradiation in UV or blue lightcuring processes

https://www.gigahertzoptik.com/enus/product/rch-16/

RCH-114

Detector head for high intensity irradiation in UV or blue lightcuring processes. Features: Separate light integrator anddetector with rigid fiber coupling, 400nm+436nm BLUE
responsivity, wide viewing angle, for the usage with optometers
and amplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/rch-17/

RCH-115

Detector head for high intensity irradiation in UV or blue lightcuring processes. Features: Separate light integrator anddetector with rigid fiber coupling, light, 436nm BLUE-Peakresponsivity, wide viewing angle, for the usage with optometers
and amplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/rch-18/

XD-9501

UV-A and UV-B detector head for use in phototherapy

https://www.gigahertzoptik.com/enus/product/xd-9501/

XD-9503

UV-A and UV-B Detector Head for use in Phototherapy

https://www.gigahertzoptik.com/enus/product/xd-9503/

XD-9506

Detector Head for use in UV Photo-biological Hazard
Measurements

https://www.gigahertzoptik.com/enus/product/xd-9506/

Gigahertz Optik GmbH

Product Image

12/17

Product Name

Description

Go to product

XD-9509

Detector head for use in UV radiation protection measurements

https://www.gigahertzoptik.com/enus/product/xd-9509/

XD-9510Detector head for use in UV radiation protection measurementsin accordance with DIN EN 12198 Safety of machinery Assessment and reduction of risks arising from radiation emitted
by machinery

https://www.gigahertzoptik.com/enus/product/xd-9510/

XD-9502

Photostability Light & UV Meter

https://www.gigahertzoptik.com/enus/product/xd-9502/

PS-3701

Detector head for plant growth

https://www.gigahertzoptik.com/enus/product/ps-3701/

PS-3702

Detector head for plant growth

https://www.gigahertzoptik.com/enus/product/ps-3702/

PS-3703

Detector head for plant growth

https://www.gigahertzoptik.com/enus/product/ps-3703/

TP-4501

Detector head for plant growth. Features: PAR, phototropism andphotomorphogenesis actinic irradiance, illuminance, for use with
X1 optometers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/tp-4501/

RW-37 with SRT-M37-L

Detector heads to measure the irradiance in W/m² and theradiance in W/(m²sr)

https://www.gigahertzoptik.com/en-us/prod
uct/rw-37usrt-m37-l/

RCH-002

Detector Head for High-Intensity Irradiation in UVA or Blue Light
Curing Processes

https://www.gigahertzoptik.com/enus/product/rch-002/

RCH-005

Detector head for high intensity irradiation in UV or blue lightcuring processes. Features: Separate light integrator anddetector with rigid fiber coupling, (320-460)nm UVABLUE
responsivity, wide viewing angle, for the usage with optometers
and amplifiers, calibration certificate.

https://www.gigahertzoptik.com/enus/product/rch-005/

Gigahertz Optik GmbH

Product Image

13/17

Product Name

Description

Go to productK-xx-CCalibration of the signal current sensitivity of optometers.
Features: calibration of all gain stages, traceable calibrated
current source, calibration certificate

https://www.gigahertzoptik.com/enus/product/k-xx-c/

ISD-5P-SiUV

Integrating sphere detector with UV-enhanced Si photodiode and
50 mm sphere for Laser power in W

https://www.gigahertzoptik.com/en-us/prod
uct/isd-5p-siuv-2/

UV-37 with SRT-M37-LUV

Detector heads to measure the UV irradiance in W/m² and the UVradiance in W/(m²sr)

https://www.gigahertzoptik.com/en-us/prod
uct/uv-37usrt-m37-luv/

UV-3726

UV detector for UV-C LEDs and low-pressure Hg germicidal lamps

https://www.gigahertzoptik.com/enus/product/uv-3726/

RCH-xxx Series

UV Detectors for measuring the UV Curing Irradiancehttps://www.gigahertzoptik.com/en-us/prod
uct/rch-xxx-series/

UV-3727

UV detector for germicidal lamps

https://www.gigahertzoptik.com/enus/product/uv-3727/

ISD-5P-IGA

Integrating sphere detector with InGaAs photodiode and 50 mm
sphere for Laser power in W

https://www.gigahertzoptik.com/en-us/prod
uct/isd-5p-iga-2/

LCR-20/LCR-21

Light-reflection hand-held meter series for flat samples, UV
reflection measurements

https://www.gigahertzoptik.com/enus/product/lcr-20/

MDC4-1-UVBLUE

Smart integral detector for UV to Blue LEDs with wavelength
detection

https://www.gigahertzoptik.com/en-us/prod
uct/mdc4-uv-blue/

ISS-28P-Xe-V01

Integrating sphere light source with very high light output (sunlike spectra)

https://www.gigahertzoptik.com/en-us/prod
uct/iss-28p-xe-v01/

MDC4-1-UV

Smart integral detector for UV LEDs with wavelength detection

https://www.gigahertzoptik.com/enus/product/mdc4-1-uv/

RCH-016

UV detector for measuring the irradiance of UV curing LEDs

https://www.gigahertzoptik.com/enus/product/rch-016/

Gigahertz Optik GmbH

Product Image

14/17

Product Name

Product Image

Description

Go to product

RCH-017

NIR detector for measuring the irradiance of NIR curing LEDs

https://www.gigahertzoptik.com/enus/product/rch-017/

RCH-117

NIR detector for measuring the irradiance of NIR curing LEDs

https://www.gigahertzoptik.com/enus/product/rch-117/

RCH-019

UV detector for measuring the irradiance of UV curing LEDs

https://www.gigahertzoptik.com/enus/product/rch-019/

RCH-119

UV detector for measuring the irradiance of UV curing LEDs

https://www.gigahertzoptik.com/enus/product/rch-119/

Silux-37xx Series

Silux detector with 37 mm housing and silux calibration for nightvision and low light level illumination rating

https://www.gigahertzoptik.com/en-us/prod
uct/silux-37xx-series/

RCH-XXXs Series

RCH-XXXs Series for accurate spherical Irradiance and fluence
rate measurements

https://www.gigahertzoptik.com/enus/product/rch-xxxs/

RCH-XXXc Series

RCH-XXXc Series for accurate cylindrical Irradiancemeasurements

https://www.gigahertzoptik.com/enus/product/rch-xxxc/

Purchasing information
Article-Nr

ModellDescription

15298890

X1-1

Meter, 2 x 1.5 V AA batteries, cable, manual. For use with -4
detectors

15309641

X1-1-V02

Optometer, 2 x 1.5 V AA batteries, cable, manual. For detector head
UV-3726-4

15311738

X1-1-V03

Optometer, 2 x 1.5 V AA batteries, cable, manual. For detector head
UV-3718-4

15312065

X1-5

Optometer, 2 x 1.5 V AA batteries, cable, manual. For use with -5
detectors.

15313179

X1-6

Optometer, 2 x 1.5 V AA batteries, cable, manual. For use with MDC4
Series detectors.

Product

Gigahertz Optik GmbH

15/17

Article-Nr

ModellDescription

K-X11-C

Current calibration at all amplification levels.
DIN EN ISO/IEC 17025 DAkkS Test Certificate in combination withdifferent detector heads can be requested optionally.

Light Detectors

Please check the light detector datasheets for specification andpurchasing information or see tab configurable with.

15298071

S-SDK-X20

For software implementation of the X20 optometer board or X1
device into custome made software. Supply of .dll's and LabView VI'sfor device communication.

15298167

S-X1

User software for the X1

15296381

X1-Z02

Adapter cable (2m) to connect light detectors with BNC (-1)
connector to the -4

15296387

X1-Z03

Adapter to connect up to four detectors with BNC connector to X1

15297973

X1-Z04Adapter cable 12 inch with ITT (-4) connector for X1. Al box with -4
socket15298036

X1-Z05Adapter cable to connect light detectors with -2 calibration data
connector to the ITT (-4) socket of the optometer X1 1. Cable length
0.2 m.

15317151

X15Z-01

Connector adapter to connect a detector with -1 (BNC) plug to an
X1-5 optometer with -5 socket.

15295292

BHO-04

Hard case for meter and accessories

15295239

BHO-05Hard case for meter and accessories

15295680

BHO-06

Hard case for meter and accessories

15297539

BHO-11

Hard case for meter and accessories

15298236

BHO-15Hard case for meter and accessories

Re-calibration
15300671

Options

SoftwareAccessories

Gigahertz Optik GmbH

16/17

Contact, Calibration, Service & Support
We are known worldwide for excellent technical consulting and after sales support. Contact us
to find together the best solution for you. Our services:
Technical Consulting & Sales
After-Sales Support
Calibrations & Re-Calibrations (ISO/IEC 17025 Calibration Services, factory calibration,
Calibration of Third-Party Products)
Repairs & Updates
OEM & Feasibility Consulting of Customized Solutions
Send us your inquiry or contact us by phone or e-mail. We would welcome your feedback too orreview us on Google.

Gigahertz Optik GmbH (Headquarter)
Tel.: +49 (0)8193-93700-0
Fax: +49 (0)8193-93700-50
info@gigahertz-optik.de
An der Kaelberweide 12
82299 Tuerkenfeld, Germany

Gigahertz-Optik, Inc. (US office)Phone: +1-978-462-1818
info-us@gigahertz-optik.com
Boston North Technology Park
Bldg B - Ste 205
Amesbury, MA 01913 USA

Gigahertz Optik GmbHPowered by TCPDF (www.tcpdf.org)

17/17