GP01163P/00/EN/02.24-00
71647907
2024-03-26

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Cerabar PMP71B
Process pressure measurement
HART

Services

Cerabar PMP71B HART

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

1.1
1.2
1.3
1.4
1.5

Document function . . . . . . . . . . . . . . . . . . . . . 4
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using this document . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . 5
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 5

2Overview of the operating menu . . . . . 6

3

Description of device parameters . . . 31

3.1
3.2
3.3
3.4
3.5

User navigation . . . . . . . . . . . . . . . . . . . . . . . 31
"Guidance" menu . . . . . . . . . . . . . . . . . . . . . . 34
"Diagnostics" menu . . . . . . . . . . . . . . . . . . . . 106"Application" menu . . . . . . . . . . . . . . . . . . . . 131"System" menu . . . . . . . . . . . . . . . . . . . . . . . 162

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 177

Endress+Hauser

3About this document

Cerabar PMP71B HART

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters. The document provides a detailed explanation of each individual parameter.
Performance of tasks that require detailed knowledge of the functioning of the device:• Commissioning measurements under difficult conditions• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Information on the document structureThis document lists the submenus and parameters that are available when the"Maintenance" option user role is enabled.
For the operating concept of the operating menus, see the Operating Instructions.

1.3.2

Structure of a parameter description

The individual parts of a parameter description are described in the following section:• Navigation: Navigation path to the parameter via the local display• Prerequisite: The parameter is only available under these specific conditions• Description: Description of the parameter function• Selection: List of the individual options for the parameter• User entry: Input range for the parameter• User interface: Display value/data of the parameter• Additional information:• On individual options• On display values/data• On the input range• On the factory setting• On the parameter function

4

Endress+Hauser

Cerabar PMP71B HART

About this document

1.4

Symbols used

1.4.1

Symbols for certain types of Information

Additional information:
Reference to documentation:
Operation via local display:
Operation via operating tool:
Write-protected parameter:

1.5

Documentation

1.5.1

Standard documentation

Operating Instructions
The Operating Instructions are available via the Internet: www.endress.com →Download

1.5.2

Supplementary device-dependent documentation

Special Documentation
The Special Documentation is available via the Internet: www.endress.com →Download

Endress+Hauser

5Overview of the operating menu

2

Cerabar PMP71B HART

Overview of the operating menu
→  31

Guidance‣ Commissioning→  34‣ Device identification
Device tag→  34

Device name→  34

Serial number→  34

Extended order code 1→  35Extended order code 2→  35Extended order code 3→  35‣ Device identification
Locking status‣ Device identification→  36→  36→  37

HART short tag→  37

HART date code→  37

HART descriptor→  37

HART message→  37

HART address→  38‣ Measurement adjustments→  38Assign PV→  38Damping→  38‣ Measurement adjustments→  39Pressure unit

→  39Temperature unit

→  39‣ Measurement adjustments6→  34→  40Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

Pressure unit

→  40Scaled variable unit

→  40Free text→  41

Temperature unit

→  41‣ Measurement adjustments
Zero adjustment→  42Pressure→  43‣ Output settings→  43
Output current transfer function‣ Output settings‣ Output settings→  43→  44Lower Range Limit→  44Upper Range Limit→  44

Minimum span→  44‣ Output settings→  45Pressure→  45

Scaled variable→  45‣ Output settings

Endress+Hauser→  43→  43

Scaled variable transfer function‣ Output settings→  42→  45Lower range value output→  45Upper range value output→  45Lower Range Limit→  46Upper Range Limit→  46

Minimum span→  46→  46

7

Overview of the operating menu

Cerabar PMP71B HART

Scaled variable transfer function→  46

Pressure value 1→  47

Scaled variable value 1→  47

Pressure value 2→  47

Scaled variable value 2→  47Lower Range Limit→  48Upper Range Limit→  48Minimum span→  48‣ Output settings→  48Lower range value output→  48Upper range value output→  49Current range output→  49Failure behavior current output→  49Failure current→  49Loop current mode→  50Assign HART variables?

→  50‣ Output settings8→  50
Process variable output current→  50Current range output→  51

Lower range value output→  51

Upper range value output→  51

Failure behavior current output→  51

Failure current→  52Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

Loop current mode→  52

Assign HART variables?

→  52‣ Output settings→  52
Assign PV→  52

Assign SV→  53

Assign TV→  53

Assign QV→  54‣ Heartbeat Technology→  114‣ Heartbeat Verification→  55‣ Heartbeat Configuration
Heartbeat Verification‣ Mainboard module→  55→  55

System status→  55

Output current→  55

Software integrity→  56RAM check→  56ROM check→  56Loop diagnostics→  57‣ Sensor module→  57
Sensor integrity→  57

Membrane integrity→  57

Sensor/membrane integrity→  58

Statistical Sensor Diagnostics→  58

Analog path integrity→  58‣ Verification result
Endress+Hauser→  55→  599Overview of the operating menu

Cerabar PMP71B HART

Verification result→  59Save protocol?

→  59‣ Finish10→  59Inspector→  59Location→  59Notes→  60

Plant operator→  60Operating time (Verification)→  60Date/time Heartbeat Verification→  60‣ SSD: Statistical Sensor Diagnostics→  60‣ Configuration→  61
Status summary→  61

Status summary→  61

Status summary→  61

Status summary→  62

Status summary→  62

Status summary→  62

Status summary→  63

Status summary→  63

Status summary→  63

System status→  64Signal status→  64Signal noise status→  64Baseline build process→  64

Sample rate

→  65Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

‣ Monitoring→  66Signal status→  66

Current Baseline signal→  66

Baseline Signal Upper Control Line→  66

Baseline Signal Control Line→  66

Baseline Signal Lower Control Line→  67Signal minimum value→  67Signal maximum value→  67‣ Monitoring→  67Signal noise status→  67

Current Baseline noise→  68Baseline Signal Noise Upper ControlLine→  68Baseline Signal Noise Control Line→  68Baseline Signal Noise Lower ControlLine→  68Baseline Signal Noise Minimum→  69Signal noise minimum value→  69Signal noise maximum value→  69‣ Diagnostic settings

Endress+Hauser→  69SSD Monitoring delay time→  69900 Event category→  69900 Event category→  69900 Diagnostic behavior→  70900 Diagnostic behavior→  70SSD Out of range delay time→  7011Overview of the operating menu

Cerabar PMP71B HART

906 Event category→  70906 Event category→  70‣ Activate/Deactivate
SSD: Statistical Sensor Diagnostics‣ Loop diagnostics→  71→  71‣ Configuration→  71
I have read the warning texts.

‣ Configuration→  71→  71Baseline status→  71Baseline build process→  72‣ Configuration→  72
Timestamp Baseline→  72

Resistance Baseline→  72

Supply voltage Baseline→  72

Timestamp previous Baseline→  73

Resistance previous Baseline→  73

Supply voltage previous Baseline→  73‣ Monitoring→  73Tolerated deviation +/-‣ Diagnostic settings12→  71→  73→  74806 Event delay→  74806 Event category→  74806 Event category→  74Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

806 Diagnostic behavior→  74806 Diagnostic behavior→  74‣ Activate/Deactivate
Loop diagnostics‣ Process window→  75→  75‣ Pressure range→  75500 Process alert pressure→  75

Low alert value→  75High alert value→  76Counter underruns of user limit Pmin

→  76Counter overruns of user limit Pmax

→  76500 Diagnostic behavior→  77500 Diagnostic behavior→  77500 Event category→  77500 Event category→  77‣ Pressure range→  77501 Process alert scaled variable→  77

Low alert value→  78High alert value→  78501 Diagnostic behavior→  78501 Diagnostic behavior→  78501 Event category→  79501 Event category→  79‣ Temperature range
User temperature process alert

Endress+Hauser→  75→  79→  7913

Overview of the operating menu

Cerabar PMP71B HART

Low alert value→  79High alert value→  79Counter underruns of user limit Tmin

→  80Counter overruns of user limit Tmax

→  80502 Diagnostic behavior→  80502 Diagnostic behavior→  80502 Event category→  80502 Event category→  80‣ Safety mode→  81‣ Preparation→  81
I have read the warning texts.

‣ Preparation→  81Enter safety unlocking code→  81

Locking status→  81‣ Preparation→  83Character test string→  83

Confirm→  83‣ Preparation‣ Preparation14→  81→  84
Device tag→  84

Device name→  84

Serial number→  85

Firmware version→  85

Hardware version→  85

Confirm→  85→  86Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

CRC device configuration→  86Stored CRC device configuration→  86

Timestamp stored CRC device config.

→  86

Operating time→  86

Configuration counter→  87‣ Confirmation→  87
Zero adjustment offset→  87

HP/LP swap

→  87

Damping→  88Sensor pressure range behavior→  88Confirm→  88‣ Confirmation→  88Output current transfer function→  88Low cutoff→  89Confirm→  89‣ Confirmation‣ Confirmation

Endress+Hauser→  89Measuring mode current output→  89Failure behavior current output→  89Current range output→  90Lower range value output→  90Upper range value output→  90Assign PV→  90Confirm→  91→  91

15

Overview of the operating menu

Cerabar PMP71B HART

Zero adjustment offset→  91

Confirm→  91‣ Locking→  91Enter safety locking code→  91

Locking status→  92

Save protocol?

→  92‣ Result→  93
Inspector→  93

Location→  93Date/time→  93Notes→  93

Plant operator→  94‣ Proof test

→  94‣ Preparation Proof test
I have read the warning texts.

‣ Preparation Proof test16

→  94→  94→  95Visual inspection→  95Notes→  95‣ Preparation Proof test

→  95

Device tag→  95

Device name→  96

Serial number→  96

Firmware version→  96

Hardware version→  97

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

‣ Preparation Proof test

→  97

CRC device configuration→  97

Stored CRC device configuration→  97

Timestamp stored CRC device config.

→  97

Operating time→  98Configuration counter→  98‣ Preparation Proof test

→  99Select test procedure→  99Safety function?

→  99‣ Simulation and check failure→  99current

Failure current (high alarm)→  99Confirm→  100(High) Is safety function triggered?

→  100

Terminal current (high alarm)→  100

Confirm→  100(Low) Is safety function triggered?

→  101‣ Proof test

Endress+Hauser→  101
Setpoint 1→  101

Pressure 1→  101

Terminal current 1→  101

Current deviation 1→  102

Setpoint 2→  102

Pressure 2→  102

17

Overview of the operating menu

Cerabar PMP71B HART

Terminal current 2→  102

Current deviation 2→  102‣ Finish→  103
Active diagnostics→  103

Summary→  103

Decision of inspector→  103

Inspector→  104

Location→  104

Date/time proof test

→  104Notes→  104

Plant operator→  105→  106

Diagnostics‣ Active diagnostics
Active diagnostics→  106

Timestamp→  106

Previous diagnostics→  107

Timestamp→  107

Operating time from restart→  107

Operating time→  108‣ Event logbook→  108
Filter options‣ Minimum/maximum values18→  106→  108→  109

Pressure min→  109

Counter limit underruns sensor Pmin

→  109

Counter underruns of user limit Pmin

→  109

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

Minimum sensor temperature→  110

Counter limit underruns sensor Tmin

→  110

Counter underruns of user limit Tmin

→  110Minimum terminal voltage→  110Minimum electronics temperature→  110

Reset user defined counters P and T→  111

Pressure max→  111

Counter limit overruns sensor Pmax

→  111

Counter overruns of user limit Pmax

→  111

Maximum sensor temperature→  111

Counter limit overruns sensor Tmax

→  112Counter overruns of user limit Tmax

→  112

Maximum terminal voltage→  112

Maximum electronics temperature→  112‣ Simulation→  113
Simulation→  113

Value pressure simulation→  113

Value current output→  113

Diagnostic event simulation→  113‣ Heartbeat Technology→  114‣ Heartbeat Verification

Endress+Hauser→  114

Date/time Heartbeat Verification→  114

Operating time (Verification)→  114

Verification result→  114

Status→  11519

Overview of the operating menu

Cerabar PMP71B HART

‣ Loop diagnostics→  115
Rebuild baseline→  115

Tolerated deviation +/-→  115

Baseline status→  116

Loop diagnostics→  116

Terminal voltage 1→  116

Clamping voltage lower threshold→  116

Clamping voltage upper threshold→  116806 Event delay→  117‣ Statistical Sensor Diagnostics
SSD: Statistical Sensor Diagnostics→  117

System status→  117Signal status→  117Signal noise status→  118‣ Diagnostic settings→  118‣ Properties20→  117→  118
SSD Out of range delay time→  118

SSD Monitoring delay time→  119500 Process alert pressure→  119

Low alert value→  119High alert value→  119501 Process alert scaled variable→  119

Low alert value→  120High alert value→  120

User temperature process alert→  120

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

Low alert value→  120High alert value→  121806 Diagnostic behavior→  121806 Diagnostic behavior→  121806 Event category→  121806 Event category→  121806 Event delay→  121‣ Configuration→  61‣ Configuration

Endress+Hauser→  122500 Diagnostic behavior→  122500 Diagnostic behavior→  122500 Diagnostic behavior→  122500 Diagnostic behavior→  122500 Event category→  122500 Event category→  122500 Event category→  122500 Event category→  122501 Diagnostic behavior→  122501 Diagnostic behavior→  122501 Diagnostic behavior→  122501 Diagnostic behavior→  122501 Event category→  123501 Event category→  123501 Event category→  123501 Event category→  123

21

Overview of the operating menu

Cerabar PMP71B HART

502 Diagnostic behavior→  123502 Diagnostic behavior→  123502 Diagnostic behavior→  123502 Diagnostic behavior→  123502 Event category→  123502 Event category→  123502 Event category→  123502 Event category→  123‣ Process22→  124806 Diagnostic behavior→  124806 Diagnostic behavior→  124806 Diagnostic behavior→  124806 Diagnostic behavior→  124806 Diagnostic behavior→  124806 Diagnostic behavior→  124806 Event category→  124806 Event category→  124806 Event category→  124806 Event category→  124806 Event category→  124806 Event category→  124822 Diagnostic behavior→  124822 Event category→  125

Sensor pressure range behavior→  125841 Event category→  125

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

→  126

900 Event category→  126

900 Event category→  126

900 Event category→  126900 Diagnostic behavior→  126900 Diagnostic behavior→  126900 Diagnostic behavior→  126900 Diagnostic behavior→  126906 Diagnostic behavior→  126906 Diagnostic behavior→  126906 Diagnostic behavior→  126906 Diagnostic behavior→  126906 Event category→  126906 Event category→  126906 Event category→  126906 Event category→  126→  131

Application‣ Measuring units

Endress+Hauser

900 Event category→  131
Pressure unit

→  131

Decimal places pressure→  131

Temperature unit

→  131

Scaled variable unit

→  132

Free text→  133

Decimal places scaled variable→  133

23

Overview of the operating menu

Cerabar PMP71B HART

‣ Measured values→  134
Pressure→  134

Scaled variable→  134

Sensor temperature→  134

Terminal voltage 1→  134

Terminal current→  135

Electronics temperature→  135‣ Sensor→  135‣ Basic settings→  135
Output current transfer function→  135

Damping→  136‣ Sensor calibration
Zero adjustment→  136

Calibration offset→  136

Zero adjustment offset→  137

Sensor Trim Reset→  137

Lower sensor trim measured value→  137

Lower sensor trim

→  138

Upper sensor trim measured value→  138

Upper sensor trim

→  139‣ Sensor limits24→  136→  139
Lower Range Limit→  139

Upper Range Limit→  139

Minimum span

→  140

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

Sensor temperature lower range limit→  140

Sensor temperature upper range limit→  140‣ Scaled variable→  140
Assign PV→  140

Scaled variable unit

→  141

Free text→  141

Pressure→  142

Scaled variable transfer function→  142

Lower range value output→  142

Upper range value output→  142

Activate table→  143

Pressure value 1→  143

Scaled variable value 1→  143

Pressure value 2→  143

Scaled variable value 2→  143

Pressure→  144

Scaled variable→  144‣ Wet calibration

Endress+Hauser→  144Zero→  144

Pressure value 1→  144Span→  144

Pressure value 2→  145Zero→  144

Lower range value output→  145

25

Overview of the operating menu

Cerabar PMP71B HARTSpan

→  144

Upper range value output→  145‣ Current output→  147
Assign PV→  147

Measuring mode current output→  147

Current range output→  147

Lower range value output→  148Upper range value output→  148

Failure behavior current output→  148

Failure current→  148Output current→  149

Terminal current→  149‣ HART output→  150‣ Configuration→  150
HART address→  150

HART short tag→  150

Device tag→  150

No. of preambles→  151

Loop current mode→  151‣ HART output

26→  151
Assign PV→  151

Primary variable (PV)→  151

Assign SV→  152

Secondary variable (SV)→  152

Assign TV→  152

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

Tertiary variable (TV)→  153

Assign QV→  153

Quaternary variable (QV)→  154‣ Burst configuration 1
Burst mode 1→  154

Burst command 1→  155

Burst variable 0→  155

Burst variable 1→  156

Burst variable 2→  156

Burst variable 3→  156

Burst variable 4→  156

Burst variable 5→  157

Burst variable 6→  157

Burst variable 7→  157

Burst trigger mode→  157

Burst trigger level→  158

Min. update period→  158

Max. update period→  158‣ Information

Endress+Hauser→  154→  159
Device ID→  159

Device type→  159

Device revision→  159

HART short tag→  160

HART revision→  160

HART descriptor→  16027

Overview of the operating menu

Cerabar PMP71B HART

HART message→  160

HART date code→  161→  162

System‣ Device management
Device tag→  162

Locking status→  162

Configuration counter→  163

Reset device→  163‣ User management28→  162→  164

User role→  164

Password→  164

Enter access code→  164

Status password entry→  165

New password→  165

Confirm new password→  165

Status password entry→  165

Old password→  165

New password→  165

Confirm new password→  165

Status password entry→  165

Reset password→  166

Status password entry→  165

Old password→  165

Status password entry→  165

Endress+Hauser

Cerabar PMP71B HART

Overview of the operating menu

‣ Bluetooth configuration
Bluetooth activation‣ Display→  166→  166

Language→  166

Format display→  167

Value 1 display→  167

Value 2 display→  168

Value 3 display→  168

Value 4 display→  169

Contrast display→  169‣ Geolocation→  170
Process Unit Tag→  170

Location Description→  170

Longitude→  170

Latitude→  170

Altitude→  171

Location method→  171‣ Information

Endress+Hauser→  166→  171
Device name→  171

Manufacturer→  172

Serial number→  172

Order code→  172

Firmware version→  173

Hardware version→  173

Extended order code 1→  17329

Overview of the operating menu

Cerabar PMP71B HART

Extended order code 2→  173

Extended order code 3→  174

Checksum→  174‣ Software configuration30→  175

CRC device configuration→  175

Stored CRC device configuration→  175

Timestamp stored CRC device config.

→  175

Activate SW option→  175

Software option overview→  176Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3

Description of device parameters

In the following section, the parameters are listed according to the menu structure of theoperating tool.
The operating menu is dynamic and adapts the choice of parameters to the selected
options.

3.1

User navigation

The Guidance main menu contains functions which enable users to perform basic tasks
swiftly, e.g. commissioning.
These are primarily guided wizards and cross-subject special functions.
Navigation

3.1.1



Guidance

Overview of the operating menu

"Guidance" menu
• Commissioning (→  34)• Heartbeat Technology (→  55)• Safety mode (→  81)• Proof test (→  94)"Diagnostics" menu
• Active diagnostics (→  106)
• Event logbook (→  108)
• Minimum/maximum values (→  109)• Simulation (→  113)
• Heartbeat Technology (→  114)
• Diagnostic settings (→  69)"Application" menu
• Measuring units (→  131)
• Measured values (→  134)
• Sensor (→  135)
• Current output (→  147)
• HART output (→  150)
"System" menu
• Device management (→  162)
• User management (→  164)
• Bluetooth configuration (→  166)
• Display (→  166)
• Geolocation (→  170)
• Information (→  159)
• Software configuration (→  175)

Endress+Hauser

31Description of device parameters

Cerabar PMP71B HART

3.1.2

Commissioning

Run this wizard to put the device into operation. Enter the appropriate value in eachparameter or select the appropriate option.
If the wizard is canceled before all the necessary parameters have been configured,
any settings already made are saved. For this reason, the device may then be in an
undefined state!
In such situations, it is advisable to reset the device to the factory default settings.
The following parameters are configured in the Commissioning wizard:• Device identification (→  34)• Device tag (→  34)• Device name (→  34)• Serial number (→  34)• Extended order code 1 (→  35)• Extended order code 2 (→  35)• Extended order code 3 (→  35)• Locking status (→  36)• HART short tag (→  37)• HART date code (→  37)• HART descriptor (→  37)• HART message (→  37)• HART address (→  38)• Measurement adjustments (→  38)• Assign PV (→  38)• Damping (→  38)• Pressure unit (→  39)• Temperature unit (→  39)• Scaled variable unit (→  40)• Zero adjustment (→  42)• Pressure (→  43)• Output settings (→  43)• Output current transfer function (→  43)• Lower Range Limit (→  44)• Upper Range Limit (→  44)• Minimum span (→  44)• Pressure (→  45)• Scaled variable (→  45)• Lower range value output (→  45)• Upper range value output (→  45)• Scaled variable transfer function (→  43)• Pressure value 1 (→  47)• Current range output (→  49)• Failure behavior current output (→  49)• Failure current (→  49)• Loop current mode (→  50)• Assign HART variables? (→  50)• Process variable output current (→  50)• Assign PV (→  38)• Assign SV (→  53)• Assign TV (→  53)• Assign QV (→  54)

32

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3.1.3

Heartbeat Technology

Heartbeat Technology offers diagnostic functionality through continuous self-monitoring,
the transmission of additional measured variables to an external Condition Monitoring
system and the in-situ verification of measuring devices in the application.
Special Documentation "Heartbeat Monitoring + Verification"
SD02525P"Heartbeat Verification" wizard
This wizard is used to start an automatic verification of the device functionality.
The results can be documented as a verification report.
"SSD: Statistical Sensor Diagnostics" wizard
Using statistical analysis of the pressure signal, process anomalies such as plugged impulse
lines can be detected. This wizard supports the settings and thresholds that should lead toa diagnostic message.
"Loop diagnostics" wizard
Using this wizard, changes in the current-voltage loop characteristics (baseline) can beused to detect unwanted installation anomalies such as creep currents caused by terminal
corrosion or a deteriorating power supply that can lead to an incorrect 4-20 mA measured
value.
"Process window" wizard
This wizard uses user-defined limits for pressure and temperature to detect unwanted
installation or application anomalies.
Applications:- Defective heat tracer or insulation- Frozen process connections- Dynamic pressure peaks etc.

3.1.4

Safety mode

The write protection guards the device settings against overwriting. In addition, it isrecommended for safety applications to confirm the safety relevant device settings. This
ensures that the correct values have been entered and downloaded to device.
This input can be used as the confirmation sequence instead of manual checklists.
After the safety relevant device settings have been confirmed, the device is marked withthe property Safety-locked. This indicates that the safety relevant parameter settings havebeen checked and evaluated as correct.
To unlock the safety locking the sequence needs to be restarted. The safety locking isdeactivated when the safety unlocking code (= safety locking code) is entered.

3.1.5

Proof testing

The proof test will simulate the current output.
The safety function is not guaranteed during proof test. Alternative process control inmanual must be taken to ensure process safety.
Note: It is only possible to perform a proof test when the device has no alarm and the
hardware write protection switch is off.

Endress+Hauser

33

Description of device parameters

Cerabar PMP71B HART

3.2

"Guidance" menu

Navigation

3.2.1



Guidance"Commissioning" wizard

Navigation

Guidance → Commissioning"Device identification" wizard
Navigation

Guidance → Commissioning → Device ident.

Device tagNavigation

Guidance → Commissioning → Device ident. → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)

Device name
Navigation

Guidance → Commissioning → Device ident. → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

Serial number



Navigation

Guidance → Commissioning → Device ident. → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

34

Max. 11-digit character string comprising letters and numbers.

Endress+Hauser

Cerabar PMP71B HART

Additional information

Description of device parameters

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

Extended order code 1Navigation

Guidance → Commissioning → Device ident. → Ext. order cd. 1

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string

Factory setting–

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.

Extended order code 2Navigation

Guidance → Commissioning → Device ident. → Ext. order cd. 2

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface

Character string

Factory setting–

Extended order code 3

Navigation

Guidance → Commissioning → Device ident. → Ext. order cd. 3

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface

Endress+Hauser

Character string35

Description of device parameters

Factory setting

Cerabar PMP71B HART–

"Device identification" wizard
Navigation

Guidance → Commissioning → Device ident.

Locking status
Navigation

Guidance → Commissioning → Device ident. → Locking status

Description

Displays the active write protection.

User interface• Hardware locked• Safety locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.
Selection
Function scope of the "Locking status" parameter

36

Options

Description

None

The access status displayed in the Access status display parameter applies. Only
appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the main electronics module.
This prevents write access to the parameters (e.g. via the local display or operatingtool).

Temporarily locked

Write access to the parameters is temporarily locked due to device-internal
processing (e.g. data upload/download, reset). Once the internal processing has
been completed, the parameters can be changed once again.

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Device identification" wizard
Navigation

Guidance → Commissioning → Device ident.

HART short tag

Navigation

Guidance → Commissioning → Device ident. → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Max. 8 characters: A to Z, 0 to 9 and certain special characters (e.g. punctuation marks, @,%).

HART date code

Navigation

Guidance → Commissioning → Device ident. → HART date code

Description

Date of the last configuration change

User entry

Character string comprising numbers, letters and special characters (10)Additional information

Date format: YYYY-MM-DD
Make sure you adhere to this format when entering the date. Otherwise errors may
occur in individual HART commands.

HART descriptor

Navigation

Guidance → Commissioning → Device ident. → HART descriptor

Description

Description for the measuring point.

User entry

Character string comprising numbers, letters and special characters (16)

HART message

NavigationDescription

A HART message which is sent via the HART protocol when requested by the master.

User entry

Character string comprising numbers, letters and special characters (32)Endress+Hauser

Guidance → Commissioning → Device ident. → HART message

37

Description of device parameters

Cerabar PMP71B HART

HART addressNavigation

Guidance → Commissioning → Device ident. → HART address

Description

Define the HART address of the device.

User entry

0 to 63

Additional information• The measured value can only be transmitted via the current value if the address is set to"0". The current is fixed at 4.0 mA for all other addresses (Multidrop mode).
• Only addresses in the range 0 to 15 are permitted for a system according to HART 5.0.
• All addresses in the range 0 to 63 are permitted for a system with HART 6.0 and higher.

"Measurement adjustments" wizard
Navigation

Guidance → Commissioning → Meas. adjust.

Assign PV

Navigation

Guidance → Commissioning → Meas. adjust. → Assign PV

Description

Use this function to select a measured variable (HART device variable) for the primary
dynamic variable (PV).

Selection• Pressure• Scaled variable

DampingNavigation

Guidance → Commissioning → Meas. adjust. → Damping

Description

The damping is effective before the measured value is further processed, i.e., before thefollowing processes:- Scaling- Limit value monitoring- Forwarding to display- Forwarding to Analog Input BlockNote:
The Analog Input Block has its own “Damping” parameter. In the measurement chain, only
one of the two attenuation parameters shall have a value other than 0.
Otherwise, the signal will be attenuated several times.

User entry

38

0 to 999.0 s

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Measurement adjustments" wizard
Navigation

Guidance → Commissioning → Meas. adjust.

Pressure unitNavigation

Guidance → Commissioning → Meas. adjust. → Pressure unit

Selection

SI units
• MPa
• kPa
• Pa
• bar
• mbar• torr• atm• kgf/cm²• gf/cm²

US unitspsi

Other units• inH2O• inH2O (4°C)• mmH2O• mmH2O (4°C)• mH2O• mH2O (4°C)• ftH2O• inHg• mmHgTemperature unitNavigationDescription

Use this function to select the unit for the temperature.

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

Selection

Endress+Hauser

Guidance → Commissioning → Meas. adjust. → Temperature unit

US units°F

39

Description of device parameters

Cerabar PMP71B HART

"Measurement adjustments" wizard
Navigation

Guidance → Commissioning → Meas. adjust.

Pressure unitNavigation

Guidance → Commissioning → Meas. adjust. → Pressure unit

Selection

SI units
• MPa
• kPa
• Pa
• bar
• mbar• torr• atm• kgf/cm²• gf/cm²

US unitspsi

Other units• inH2O• inH2O (4°C)• mmH2O• mmH2O (4°C)• mH2O• mH2O (4°C)• ftH2O• inHg• mmHgScaled variable unitNavigationDescription

Use "Free text", first selection, if the desired unit is not available in the selection list. It is
possible to define a customer specific unit with another parameter.

40

Guidance → Commissioning → Meas. adjust. → Scaled Unit

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters

SI units
•%
• mm
• cm
•m
•l
• hl
• m³
•g
• kg
•t
• g/s
• kg/s• kg/min• kg/h• t/min• t/h• t/d• m³/s• m³/min• m³/h• m³/d• l/s• l/min• l/h• Nm³/h• Nl/h• Sm³/s• Sm³/min• Sm³/h• Sm³/d• Nm³/s• g/cm³• kg/m³• Nm³/min• Nm³/dUS units
• ft
• in
• ft³
• gal (us)• bbl (us;oil)
• oz
• lb
• STon• lb/s• lb/min• lb/h• STon/min• STon/h• STon/d• ft³/s• ft³/min• ft³/h• ft³/d• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• Sft³/min• Sft³/h• Sft³/d

Imperial units• gal (imp)• gal/s (imp)• gal/min (imp)• gal/h (imp)

Custom-specific units
Free text

Free text

Navigation

Guidance → Commissioning → Meas. adjust. → Free text

User entry

Character string comprising numbers, letters and special characters (32)Temperature unitNavigationDescription

Use this function to select the unit for the temperature.

Endress+Hauser

Guidance → Commissioning → Meas. adjust. → Temperature unit

41Description of device parameters

Cerabar PMP71B HART

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

Selection

US units°F"Measurement adjustments" wizard
Navigation

Guidance → Commissioning → Meas. adjust.

Zero adjustment

NavigationDescription

Due to the mounting position of the measuring instrument, a pressure shift may occur.
The pressure shift can be corrected with the zero adjustment.

Selection• No• Confirm

42

Guidance → Commissioning → Meas. adjust. → Zero adjustment

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Pressure
Navigation

Guidance → Commissioning → Meas. adjust. → Pressure"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Output current transfer function
Navigation

Guidance → Commissioning → Output settings → Curr. trans.func

Description

Linear
The linear pressure signal is used for the current output. The flow must be calculated in
the evaluation unit.
Square root - differential pressure only
The root flow signal is used for the current output. The 'Flow (square root)'current signal is indicated on the on-site display with a root symbol.

User interface• Linear• Square root *"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Scaled variable transfer function



Navigation

Guidance → Commissioning → Output settings → Scaled function

Description"Linear'
The linear pressure signal is used for the output signal. The flow must be calculated in theevaluation unit.
"Square root" (Deltabar)
The root flow signal is used for the output signal. The "Flow (square root)'
output signal is indicated on the on-site display with a root symbol.
"Table'The output is defined according to the scaled variable / pressure table entered.

*

Visibility depends on order options or device settings

Endress+Hauser

43Description of device parameters

Selection

Cerabar PMP71B HART

• Linear• Square root *• Table"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Lower Range Limit
Navigation

Guidance → Commissioning → Output settings → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
Navigation

Guidance → Commissioning → Output settings → URL

Description

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

Minimum span
NavigationDescription

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number*

44

Guidance → Commissioning → Output settings → Minimum span

Visibility depends on order options or device settings

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Pressure

Navigation

Guidance → Commissioning → Output settings → Pressure

User entry

Signed floating-point number

Scaled variable

Navigation

Guidance → Commissioning → Output settings → Scaled variable

User entry

Signed floating-point number"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Lower range value output

Navigation

Guidance → Commissioning → Output settings → Low.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Upper range value output

NavigationDescription

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Endress+Hauser

Guidance → Commissioning → Output settings → Upp.range outp

45Description of device parameters

Cerabar PMP71B HART

Lower Range Limit
Navigation

Guidance → Commissioning → Output settings → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
Navigation

Guidance → Commissioning → Output settings → URL

Description

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

Minimum span
Navigation

Guidance → Commissioning → Output settings → Minimum span

Description

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Scaled variable transfer function



Navigation

Guidance → Commissioning → Output settings → Scaled function

Description"Linear'
The linear pressure signal is used for the output signal. The flow must be calculated in theevaluation unit.
"Square root" (Deltabar)
The root flow signal is used for the output signal. The "Flow (square root)'
output signal is indicated on the on-site display with a root symbol.
"Table'The output is defined according to the scaled variable / pressure table entered.

46

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters• Linear• Square root *• TablePressure value 1Navigation

Guidance → Commissioning → Output settings → Pressure 1

Description

Enter pressure for the first scaling point. "Scaled variable value 1" will be allocated to thispressure.

User entry

Signed floating-point number

Scaled variable value 1
Navigation

Guidance → Commissioning → Output settings → Scaled 1

Description

Enter value for the first scaling point. This value is allocated to "Pressure value 1".

User interface

Signed floating-point number

Pressure value 2Navigation

Guidance → Commissioning → Output settings → Pressure 2

Description

Enter pressure for the second scaling point. "Scaled variable value 2" will be allocated tothis pressure.

User entry

Signed floating-point number

Scaled variable value 2NavigationDescription

Enter value for the second scaling point. This value is allocated to "Pressure value 2".

User entry

Signed floating-point number*

Guidance → Commissioning → Output settings → Scaled 2

Visibility depends on order options or device settings

Endress+Hauser

47

Description of device parameters

Cerabar PMP71B HART

Lower Range Limit
Navigation

Guidance → Commissioning → Output settings → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
Navigation

Guidance → Commissioning → Output settings → URL

Description

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

Minimum span
Navigation

Guidance → Commissioning → Output settings → Minimum span

Description

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Lower range value output

NavigationDescription

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

48

Guidance → Commissioning → Output settings → Low.range outp

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Upper range value output

Navigation

Guidance → Commissioning → Output settings → Upp.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Current range output

Navigation

Guidance → Commissioning → Output settings → Current range

Description

Defines the current range used to transmit the measured or calculated value.
In brackets are indicated the “low saturation value” and the “high saturation value”.
If Measured value ≤ “low saturation”, the output current is set to “low saturation”.
If Measured value ≥ “high saturation”, the output current is set to ”high saturation”.
Note:
Currents below 3.6 mA or above 21.5 mA can be used to signal an alarm.

Selection• 4...20 mA (4...20.5 mA)• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)

Failure behavior current output

Navigation

Guidance → Commissioning → Output settings → Failure behav.

Description

Defines which current the output assumes in the case of an error.
Min: < 3.6 mA
Max: >21.5 mA
Note: The hardware DIP Switch for alarm current has priority over software setting.

Selection• Min.
• Max.

Failure current

NavigationDescription

Enter current output value in alarm condition

User entry

21.5 to 23 mA

Endress+Hauser

Guidance → Commissioning → Output settings → Failure current

49Description of device parameters

Cerabar PMP71B HART

Loop current mode
Navigation

Guidance → Commissioning → Output settings → Loop curr mode

Description

If Loop current mode is disabled, Multi-drop communication mode is activated. Multi-drop
is a HART digital communication mode where multiple devices may share the same pair ofwires for power and communications.
In this mode the output current is fixed.

User interface• Disable• Enable

Assign HART variables?
Navigation

Guidance → Commissioning → Output settings → Assign HART var?

Description

Up to four HART variables can be transmitted via the HART protocol.
Select "Yes" to show/assign measuring variables to these HART variables.

Selection

• No
• Yes

"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Process variable output current
NavigationDescription

Determines which process variable is transmitted via the current output.

User interface• Pressure• Scaled variable50

Guidance → Commissioning → Output settings → Proc.var.curr.

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Current range output

Navigation

Guidance → Commissioning → Output settings → Current range

Description

Defines the current range used to transmit the measured or calculated value.
In brackets are indicated the “low saturation value” and the “high saturation value”.
If Measured value ≤ “low saturation”, the output current is set to “low saturation”.
If Measured value ≥ “high saturation”, the output current is set to ”high saturation”.
Note:
Currents below 3.6 mA or above 21.5 mA can be used to signal an alarm.

Selection• 4...20 mA (4...20.5 mA)• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)

Lower range value output

Navigation

Guidance → Commissioning → Output settings → Low.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Upper range value output

Navigation

Guidance → Commissioning → Output settings → Upp.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Failure behavior current output

Navigation

Guidance → Commissioning → Output settings → Failure behav.

Description

Defines which current the output assumes in the case of an error.
Min: < 3.6 mA
Max: >21.5 mA
Note: The hardware DIP Switch for alarm current has priority over software setting.

Selection

Endress+Hauser• Min.
• Max.

51

Description of device parameters

Cerabar PMP71B HART

Failure current

Navigation

Guidance → Commissioning → Output settings → Failure current

Description

Enter current output value in alarm condition

User entry

21.5 to 23 mA

Loop current mode
Navigation

Guidance → Commissioning → Output settings → Loop curr mode

Description

If Loop current mode is disabled, Multi-drop communication mode is activated. Multi-drop
is a HART digital communication mode where multiple devices may share the same pair ofwires for power and communications.
In this mode the output current is fixed.

User interface• Disable• Enable

Assign HART variables?
Navigation

Guidance → Commissioning → Output settings → Assign HART var?

Description

Up to four HART variables can be transmitted via the HART protocol.
Select "Yes" to show/assign measuring variables to these HART variables.

Selection

• No
• Yes

"Output settings" wizard
Navigation

Guidance → Commissioning → Output settings

Assign PV

NavigationDescription

Use this function to select a measured variable (HART device variable) for the primary
dynamic variable (PV).

52

Guidance → Commissioning → Output settings → Assign PV

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters• Pressure• Scaled variable

Assign SVNavigation

Guidance → Commissioning → Output settings → Assign SV

Description

Use this function to select a measured variable (HART device variable) for the secondary
dynamic variable (SV).

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Terminal current *• Terminal voltage *• Median of pressure signal *• Noise of pressure signal *• Signal noise detected *• Percent of range• Loop current• Not used

Additional information

Selection• Sensor pressure option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.
• Terminal current option
The terminal current is the read-back current on terminal block.
• Signal noise detected option
0 % - Signal noise is within the permissible range.
100 % - Signal noise is outside of the permissible range.
• Loop current option
The loop current is the output current set by the applied pressure.

Assign TV

NavigationDescription

Use this function to select a measured variable (HART device variable) for the tertiary(third) dynamic variable (TV).

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Terminal current **Guidance → Commissioning → Output settings → Assign TV

Visibility depends on order options or device settings

Endress+Hauser

53Description of device parameters

Cerabar PMP71B HART

• Terminal voltage *• Median of pressure signal *• Noise of pressure signal *• Signal noise detected *• Percent of range• Loop current• Not used
Additional information

Selection• Sensor pressure option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.
• Terminal current option
The terminal current is the read-back current on terminal block.
• Signal noise detected option
0 % - Signal noise is within the permissible range.
100 % - Signal noise is outside of the permissible range.
• Loop current option
The loop current is the output current set by the applied pressure.

Assign QVNavigation

Guidance → Commissioning → Output settings → Assign QV

Description

Use this function to select a measured variable (HART device variable) for the quaternary(fourth) dynamic variable (QV).

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Terminal current *• Terminal voltage *• Median of pressure signal *• Noise of pressure signal *• Signal noise detected *• Percent of range• Loop current• Not used

Additional information

Selection• Sensor pressure option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.
• Terminal current option
The terminal current is the read-back current on terminal block.
• Signal noise detected option
0 % - Signal noise is within the permissible range.
100 % - Signal noise is outside of the permissible range.
• Loop current option
The loop current is the output current set by the applied pressure.

*

54

Visibility depends on order options or device settings

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3.2.2"Heartbeat Technology" submenu"Heartbeat Verification" wizard
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif.

Heartbeat Verification
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Heartbeat Config → Heartbeat
Verif.

Selection• Start verification• Show results"Mainboard module" wizard
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard
moduleSystem status
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard module → SystemstatusDescription

Checks active measurement device errors at diagnostical behavior "alarm".
If an active error is detected, then verification will be performed but the overall result will
always be "Failed".

User interface• Not done
• Passed• Not done
• Failed

Output current
NavigationDescription

Checks whether read-back current at the output matches the current set by the device.

Endress+Hauser

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard module → Output
curr.

55

Description of device parameters

User interface

Cerabar PMP71B HART

• Not done
• Passed• Not done
• Failed

Software integrity
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard module → Software
integ.

Description

Checks whether the function blocks of the software are executed in the correct order.

User interface• Not done
• Passed• Not done
• Failed

RAM check
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard module → RAM
check

Description

Checks the correct function of the RAM (Random Access Memory).

User interface• Not done
• Passed• Not done
• Failed

ROM check
NavigationDescription

Checks the correct function of the ROM memory (Read-Only-Memory).

User interface• Not done
• Passed• Not done
• Failed

56

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard module → ROM
check

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Loop diagnostics
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Mainboard module → Loopdiagn.

Description

Checks whether the voltage/current values are within range or the voltage/current
baseline defined with wizard.
"Failed" can indicate faulty power supply or grounding / wiring.

User interface• Not done
• Passed• Not done
• Failed

Sensor integrity
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Sensor module → Sensor
integrity

Description

Checks the integrity of the sensor. Scope of check depends on sensor type used.

User interface• Not done
• Passed• Not done
• Failed

Membrane integrity
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Sensor module → Membrane
integr.

Description

Checks the integrity of the membrane.
Notice:Not included in the scope of testing for metallic membranes.

User interface

Endress+Hauser• Not done
• Passed• Not done
• Failed57

Description of device parameters

Cerabar PMP71B HART

Sensor/membrane integrity
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Sensor module → Sensor/
membrane

Description

Checks the integrity of the sensor and membrane.

User interface• Not done
• Passed• Not done
• Failed

Statistical Sensor Diagnostics
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Sensor module → SSD

Description

Checks whether the actual values are within the defined signal noise thresholds.

User interface• Not done
• Passed• Not done
• Failed

Analog path integrity
NavigationDescription

Checks if the analog path of the sensor has changed from state of delivery.

User interface• Not done
• Passed• Not done
• Failed

58

Guidance → Heartbeat Techn. → Heartbeat Verif. → Sensor module → Path integrity

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Verification result
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Verific. result → Verific. result

User interface• Not done
• Passed• Not done
• Failed

Save protocol?
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Verific. result → Save protocol?

Description

The Report can be saved for archiving.

Selection

• No
• Yes

Inspector
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Finish → Inspector

Description

The entered inspector name will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)

Location
NavigationDescription

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)Endress+Hauser

Guidance → Heartbeat Techn. → Heartbeat Verif. → Finish → Location

59

Description of device parameters

Cerabar PMP71B HART

Notes
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Finish → Notes

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (255)

Plant operator
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Finish → Plant operator

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)

Operating time (Verification)
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Finish → Operating time

User interface

Days (d), hours (h), minutes (m), seconds (s)Date/time Heartbeat Verification
Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif. → Finish → Date/time Heartbeat
Verification

Description

Date and time of last Heartbeat Verification.
This value is updated with every Heartbeat verification.
Note:
If time information is not available, e.g. Heartbeat verification is started from display,'--------' is shown.

User interface

60

Character string comprising numbers, letters and special characters

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

Activate SSD.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

Please wait. Function is not ready.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

The signal noise is too small for the teach-in phase.
Note:
Teach-in is only possible while the process is running.
Measures:Check valve position.
If necessary, perform the teach-in procedure later while the process is running.

User interface

Endress+Hauser• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline61Description of device parameters

Cerabar PMP71B HART

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

The teach-in phase is completed. Continue or terminate the configuration.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

The SSD is not active because the average value of the raw signal is outside the limits.
The SSD is reactivated as soon as the average value is within the limits again.
Adjust the limits if necessary.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Status summary
NavigationDescription

The SSD has detected an event (e.g. blocked impulse line).
Check whether maintenance work is required.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

62

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

The SSD is not active because the process conditions are too dynamic for reliable operation.
If necessary, teach in a new baseline or adjust the sampling rate.
The SSD is reactivated as soon as the average value of the raw signal is within the limits.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Status summary
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary

Description

The current signal noise is too small to activate the SSD.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Status summary
NavigationDescription

Baseline could not be created with following reasons:- There is not enough signal noise during building baseline phase.
- There is a process change during building baseline phase.

User interface• Deactivated• Learning phase• Monitoring inactive• Monitoring active• Monitoring active with event• No baseline

Endress+Hauser

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Status summary63

Description of device parameters

Cerabar PMP71B HART

System status
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → System status

User interface• Idle• No sufficient signal noise• Stable• Not stable• Verify System Dynamics• Process dynamic too highSignal status
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Signal status

User interface• Idle• Building Baseline• Verifying Baseline• Verifying baseline failed• Monitoring• Out of range• Monitoring inactive

Signal noise status
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Noise status

User interface• Idle• Building Baseline• Verifying Baseline• Verifying baseline failed• Monitoring• Out of range• Monitoring inactive

Baseline build process
Navigation

User interface

0 to 100 %64

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Baseline process

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Sample rate

Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Sample rate

Description

Determines the sampling rate depending on the process conditions:"Fast'
homogeneous stable process with Gauss distribution.
"Medium'
dynamic process"Slow'
extremely dynamic variable process

Selection• Fast• Medium• Slow

Counter Baseline creation SSD
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Counter
Baseline

Description

Specifies how often the baseline has been rebuilt.

User interface

Positive integer

Baseline is available
NavigationDescription

Indicates whether a baseline already exists.

User interface• Please select
• No
• Yes

Endress+Hauser

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Configuration → Baseline avail.

65Description of device parameters

Cerabar PMP71B HART

Signal status
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Signal status

User interface• Idle• Building Baseline• Verifying Baseline• Verifying baseline failed• Monitoring• Out of range• Monitoring inactive

Current Baseline signal
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Curr. Baseline

Description

Current average of the raw signal.

User interface

Signed floating-point number

Baseline Signal Upper Control LineNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline S. UCL

Description

Upper limit for the average value of the raw signal. If the average value is above this limit,
the SSD is inactive.
Note:
This parameter should not be greater than "Signal maximum value".

User entry

Signed floating-point number

Baseline Signal Control Line
NavigationDescription

Learned-in mean of the raw signal.

User interface

Signed floating-point number

66

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline S. CL

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Baseline Signal Lower Control LineNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline S. LCL

Description

Lower limit for the average value of the raw signal. If the average value is below this limit,
the SSD is inactive.
Note:
This parameter should not be less than "Signal minimum value".

User entry

Signed floating-point number

Signal minimum value
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Minimum value

Description

Minimum mean of the raw signal during the learning phase.

User interface

Signed floating-point number

Signal maximum value
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Maximum value

Description

Maximum mean of the raw signal during the learning phase.

User interface

Signed floating-point number

Signal noise status
Navigation

User interface• Idle• Building Baseline• Verifying Baseline• Verifying baseline failed• Monitoring• Out of range• Monitoring inactive

Endress+Hauser

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Noise status67

Description of device parameters

Cerabar PMP71B HART

Current Baseline noise
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Current noise

Description

Current signal noise (standard deviation) of the raw signal.

User interface

Signed floating-point number

Baseline Signal Noise Upper Control LineNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline SN. UCL

Description

Upper limit for the noise of the raw signal. If the noise is above this limit, the SSD isinactive.
Note:
This parameter should not be greater than "Signal noise maximum value".

User entry

Signed floating-point number

Baseline Signal Noise Control Line
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline SN. CL

Description

Learned-in noise of the raw signal.

User interface

Signed floating-point number

Baseline Signal Noise Lower Control LineNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline SN. LCL

Description

Lower limit for the noise of the raw signal. If the noise is below this limit, the SSD isinactive.
Note:
This parameter should not be less than "Signal noise minimum value".

User entry

68

Signed floating-point number

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Baseline Signal Noise MinimumNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Baseline SN. Min

Description

Minimum value of the signal noise.
Below this value, the SSD cannot be activated.

User entry

Signed floating-point number

Signal noise minimum value
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Min. noise value

Description

Minimum measured signal noise during the learning phase.

User interface

Signed floating-point number

Signal noise maximum value
Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Monitoring → Max noise value

Description

Maximum measured signal noise during the learning phase.

User interface

Signed floating-point number

SSD Monitoring delay timeNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Diag. settings → SSD Verz. Zeit

User entry

0 to 86 400 s

900 Event category



Navigation

 Guidance → Heartbeat Techn. → Stat. Sens. Diag → Diag. settings → 900Event
category

Description

Select category for diagnostic message.

Endress+Hauser

69Description of device parameters

Selection

Cerabar PMP71B HART

• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)900 Diagnostic behavior

Navigation

 Guidance → Heartbeat Techn. → Stat. Sens. Diag → Diag. settings → 900 Diag.
behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

SSD Out of range delay timeNavigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Diag. settings → SSD Delay time

User entry

0 to 604 800 s

906 Event category



Navigation

 Guidance → Heartbeat Techn. → Stat. Sens. Diag → Diag. settings → 906Event
category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)70

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

SSD: Statistical Sensor Diagnostics



Navigation

Guidance → Heartbeat Techn. → Stat. Sens. Diag → Activ./Deactiv. → Stat. Sens. Diag

Description

Enable or disable SSD.
After selecting "Disable", no statistical sensor diagnosis takes place. No diagnostic messages
are output.

Selection• Disable• Enable

I have read the warning texts.



Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → warning texts

Description

The characterization of the electrical loop requires intermediate generation of a current
output of 4 mA and 20 mA.
Therefore, do not use the device in an active current loop during initialization of the Signal
Loop Diagnostics.

Selection

Yes

Baseline status
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Baseline status

Description"Failed"Means, baseline is not available or creation not possible.
"Success"
Baseline is available.

User interface

Endress+Hauser• Failed• Success71

Description of device parameters

Cerabar PMP71B HART

Baseline build process
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Baseline process

User interface

0 to 100 %Timestamp Baseline
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Timestamp BL

Description

Time stamp when the baseline was recorded (current-voltage curve).
At this time, the upper terminal voltage (U1) was determined at 4 mA and the lower
terminal voltage(U2) was determined at 20 mA.

User interface

Character string comprising numbers, letters and special characters

Resistance Baseline
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Resistance BL

Description

Displays the resistance value of the baseline (slope of the current-voltage curve).

User interface

Positive floating-point number

Supply voltage Baseline
NavigationDescription

Displays the point from which the baseline is built (voltage at the point where the currentvoltage curve is generated).

User interface

0.0 to 50.0 V

72

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Supply volt. BL

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Timestamp previous Baseline
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Timestamp pre BL

Description

Time stamp when the previous baseline was recorded (current-voltage curve).
At this time, the upper terminal voltage (U1) was determined at 4 mA and the lower
terminal voltage(U2) was determined at 20 mA.

User interface

Character string comprising numbers, letters and special characters

Resistance previous Baseline
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Resist. pre. BL

Description

Displays the resistance value of the previous baseline (slope of the previous currentvoltage curve).

User interface

Positive floating-point number

Supply voltage previous Baseline
Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Configuration → Voltage pre. BL

Description

Displays the point from which the previous baseline is built (voltage at the point where theprevious current-voltage curve was generated).

User interface

0.0 to 50.0 V

Tolerated deviation +/-Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Monitoring → Toler. deviation

Description

A value should be chosen to ensure that normal voltage deviations do not lead tounwanted messages.
Default
1.5 V DC

User entry

Endress+Hauser

0.5 to 3.0 V

73

Description of device parameters

Cerabar PMP71B HART

806 Event delay

Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Diag. settings → 806 Event delay

Description

Displays how long the triggering status must be present until an event message is issued.
Used to filter out short-term signal interference.

User entry

0 to 60 s

806 Event category



Navigation

 Guidance → Heartbeat Techn. → Loop diagn. → Diag. settings → 806Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)806 Diagnostic behavior

Navigation

 Guidance → Heartbeat Techn. → Loop diagn. → Diag. settings → 806 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection

74• Warning• Logbook entry only

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Loop diagnostics



Navigation

Guidance → Heartbeat Techn. → Loop diagn. → Activ./Deactiv. → Loop diagn.

Description

Enable/disable loop diagnostics.
Note:
If the function is disabled, there is no analysis and no event message.

Selection• Disable• Enable

500 Process alert pressureNavigation

Guidance → Heartbeat Techn. → Process window → Pressure range → 500 Pressure

Description

Define whether user-defined pressure limits should be set.
If "Off" is selected, no analysis will take place and no event message will be generated.

Selection

• Off
• On

Low alert value

NavigationDescription

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

Endress+Hauser

Guidance → Heartbeat Techn. → Process window → Pressure range → Low alert
value

75

Description of device parameters

Cerabar PMP71B HART

High alert value

Navigation

Guidance → Heartbeat Techn. → Process window → Pressure range → High alert
value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

Counter underruns of user limit Pmin
Navigation

Guidance → Heartbeat Techn. → Process window → Pressure range → Counter < P
user

Description

Counts how many times the value underruns the minimum values defined by the user.
User defined minimum values are shown in Diagnostic/Diagnostic settings/Properties
menu.

User interface

0 to 65 535Counter overruns of user limit Pmax
NavigationDescription

Counts how many times the value overruns the maximum values defined by the user.
User defined maximum values are shown in Diagnostic/Diagnostic settings/Properties
menu.

User interface

0 to 65 535

76

Guidance → Heartbeat Techn. → Process window → Pressure range → Counter > P
user

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

500 Diagnostic behavior

Navigation

 Guidance → Heartbeat Techn. → Process window → Pressure range → 500 Diag.
behav.

Description

Select event behavior"Logbook entry only":
no digital or analog transmission of the message"Warning": Current output unchanged. Message is output digitally (default).
"Alarm": Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

500 Event category



Navigation

 Guidance → Heartbeat Techn. → Process window → Pressure range → 500Event
category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)501 Process alert scaled variable

NavigationDescription

Define whether user-defined limits should be set.
If "Off" is selected, no analysis will take place and no event message will be generated.

Selection

• Off
• On

Endress+Hauser

Guidance → Heartbeat Techn. → Process window → Pressure range → 501 Scaled
var.

77

Description of device parameters

Cerabar PMP71B HART

Low alert value

Navigation

Guidance → Heartbeat Techn. → Process window → Pressure range → Low alert
value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

High alert value

Navigation

Guidance → Heartbeat Techn. → Process window → Pressure range → High alert
value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

501 Diagnostic behavior

Navigation

 Guidance → Heartbeat Techn. → Process window → Pressure range → 501 Diag.
behav.

Description

Select event behavior"Logbook entry only":
no digital or analog transmission of the message"Warning": Current output unchanged. Message is output digitally (default).
"Alarm": Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection

78

• Off
• Alarm• Warning• Logbook entry only

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

501 Event category



Navigation

 Guidance → Heartbeat Techn. → Process window → Pressure range → 501Event
category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)User temperature process alert

Navigation

Guidance → Heartbeat Techn. → Process window → Temp. range → UserTemp alert

Description

Define whether the user-defined sensor temperature limits should be set. If "Off" noanalysis and therefore no event message will take place.

Selection

• Off
• On

Low alert value

Navigation

Guidance → Heartbeat Techn. → Process window → Temp. range → Low alert value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry–50 to 150 °C

High alert value

NavigationDescription

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry–50 to 150 °C

Endress+Hauser

Guidance → Heartbeat Techn. → Process window → Temp. range → High alert value

79

Description of device parameters

Cerabar PMP71B HARTCounter underruns of user limit Tmin
Navigation

Guidance → Heartbeat Techn. → Process window → Temp. range → Counter < T user

User interface

0 to 65 535Counter overruns of user limit Tmax
Navigation

Guidance → Heartbeat Techn. → Process window → Temp. range → Counter > T user

User interface

0 to 65 535

502 Diagnostic behavior

Navigation

 Guidance → Heartbeat Techn. → Process window → Temp. range → 502 Diag. behav.

Description

Select event behavior"Logbook entry only":
no digital or analog transmission of the message"Warning": Current output unchanged. Message is output digitally (default).
"Alarm": Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

502 Event category



Navigation

 Guidance → Heartbeat Techn. → Process window → Temp. range → 502Event
category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)80

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3.2.3"Safety mode" wizard

Navigation

Guidance → Safety mode

"Preparation" wizard
Navigation

Guidance → Safety mode → Preparation

I have read the warning texts.
Navigation

Guidance → Safety mode → Preparation → warning texts

Description- Prior to using the wizard, read the Functional Safety Manual or the WHG documentation.
- In the wizard, safety-relevant parameter settings of the device will be shown.
- The user must confirm the correctness of these settings.
- By clicking "Finish" at the end of the wizard, the device is safety locked.
- A documentation of the safety-relevant parameter settings can be generated at the endof the wizard.
- The device can be unlocked via this wizard in case the device is safety locked.
- If the wizard is aborted, the device locking status will not be changed!

Selection

Yes"Preparation" wizard
Navigation

Guidance → Safety mode → Preparation

Enter safety unlocking code

Navigation

Guidance → Safety mode → Preparation → Safe.unlock code

Description

The Safety locking/unlocking code can be found in the corresponding safety manual or the
WHG documentation.

User entry

0 to 65 535

Locking status
NavigationDescription

Displays the active write protection.

Endress+Hauser

Guidance → Safety mode → Preparation → Locking status81

Description of device parameters

Cerabar PMP71B HART

User interface• Hardware locked• Safety locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.
Selection
Function scope of the "Locking status" parameter
Options

Description

None

The access status displayed in the Access status display parameter applies. Only
appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the main electronics module.
This prevents write access to the parameters (e.g. via the local display or operatingtool).

Temporarily locked

Write access to the parameters is temporarily locked due to device-internal
processing (e.g. data upload/download, reset). Once the internal processing has
been completed, the parameters can be changed once again.

"Preparation" wizard
Navigation

Guidance → Safety mode → Preparation

Proof test via Bluetooth allowed?



Navigation

Guidance → Safety lock → Preparation → BLE proof test

Description

After completion of the Safety mode wizard, the device will be write protected via softwarelock.
To use the proof test wizard (optional), the device does not have to be unlocked.
It must be defined, if the proof test wizard via Bluetooth is allowed.

Selection

82

• No
• Yes

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Preparation" wizard
Navigation

Guidance → Safety lock → Preparation

Enter safety locking code

Navigation

Guidance → Safety lock → Preparation → Safety code

Description

Enter the locking code to start the SIL/WHG locking sequence.

User entry

0 to 65 535

Additional information

Locking codes• WHG: 7450
• SIL: 7452
• SIL and WHG: 7454

"Preparation" wizard
Navigation

Guidance → Safety mode → Preparation

Character test string
Navigation

Guidance → Safety mode → Preparation → Char.test string

Description

The following character string is displayed:
0123456789+-,.
Set the "Confirm" parameter to "Yes" if this string is rendered correctly.
Set the "Confirm" parameter to "No" if this string is not rendered correctly. Safety locking is
not possible in this case.

User interface

Character string comprising numbers, letters and special characters

ConfirmNavigation

Guidance → Safety mode → Preparation → Confirm

Description

The following character string is displayed:
0123456789+-,.
Set the "Confirm" parameter to "Yes" if this string is rendered correctly.
Set the "Confirm" parameter to "No" if this string is not rendered correctly. Safety locking is
not possible in this case.

Endress+Hauser

83Description of device parameters

Selection

Cerabar PMP71B HART

• No
• Yes

"Preparation" wizard
Navigation

Guidance → Safety mode → Preparation

Device tag
Navigation

Guidance → Safety mode → Preparation → Device tag

Description

Displays a unique name for the measuring point so it can be identified quickly within theplant. The name is displayed in the header.

User interface

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Additional information

User interface
1

XXXXXXXXX

A0029422

1

Position of the header text on the display

The number of characters displayed depends on the characters used.

Device name
NavigationDescription

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

84

Guidance → Safety mode → Preparation → Device name

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Serial number



Navigation

Guidance → Safety mode → Preparation → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

Firmware version
Navigation

Guidance → Safety mode → Preparation → Firmware version

Description

Displays the device firmware version that is installed.

User interface

Character string in the format xx.yy.zz

Additional information

User interface
The Firmware version is also located:• On the title page of the Operating instructions• On the transmitter nameplate

Hardware version
Navigation

Guidance → Safety mode → Preparation → Hardware version

User interface

Character string comprising numbers, letters and special characters

Confirm
NavigationDescription

Confirm connection to the correct device.

Selection

• No
• Yes

Endress+Hauser

Guidance → Safety mode → Preparation → Confirm

85Description of device parameters

Cerabar PMP71B HART

"Preparation" wizard
Navigation

Guidance → Safety mode → Preparation

CRC device configuration
Navigation

Guidance → Safety mode → Preparation → CRC device conf.

Description

CRC device configuration based on current settings of safety relevant parameters.
The CRC device configuration is unique and can be used to detect changes in safety
relevant parameter settings.

User interface

0 to 65 535

Stored CRC device configuration
Navigation

Guidance → Safety mode → Preparation → Stored CRC conf.

Description

Stored CRC after the last safety lock. Factory delivery is 65535 means that the device has
not yet been safety locked.

User interface

0 to 65 535

Timestamp stored CRC device config.
Navigation

Guidance → Safety mode → Preparation → Time stored CRC

Description

Gives the time stamp when the CRC was last stored following completion of the safety lockwizard.

User interface

Character string comprising numbers, letters and special characters

Operating time
NavigationDescription

Indicates how long the device has been in operation.

Additional information

Maximum time: 9 999 d (≈ 27 years)

86

Guidance → Safety mode → Preparation → Operating time

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Configuration counter
Navigation

Guidance → Safety mode → Preparation → Config. counter

Description

Displays the counter for changes to the device parameters.
Additional information:- If the value for a static parameter is changed when optimizing or configuring theparameter, the counter is incremented by 1. This is to enable tracking different parameterversions.
- When multiple parameters are changed simultaneously, e.g. when loading parameters
into the device from an external source such as FieldCare, the counter may display a higher
value. The counter cannot be reset, nor is it reset to a default value on performing a device
reset.
- Once the counter has reached the value 65535, it restarts at 0.

User interface

0 to 65 535"Confirmation" wizard
Navigation

Guidance → Safety mode → Confirmation

Zero adjustment offset
Navigation

Guidance → Safety mode → Confirmation → Zero offset



Guidance → Safety mode → Confirmation → Zero offset

Description

Assigned value of zero adjustment due to mounting position.

User interface

Character string comprising numbers, letters and special characters

HP/LP swap
NavigationDescription

Assigned setting high pressure / low pressure.

User interface

• No
• Yes

Endress+Hauser

Guidance → Safety mode → Confirmation → HP/LP swap

87Description of device parameters

Cerabar PMP71B HART

Damping
Navigation

Guidance → Safety mode → Confirmation → Damping

Description

Assigned damping value.

User interface

Character string comprising numbers, letters and special characters

Sensor pressure range behavior
Navigation

Guidance → Safety mode → Confirmation → P-range behavior

Description

Assigned event behavior in case of over/under pressure outside of measuring range.

User interface• Alarm• Warning• Remark• Special

Confirm
Navigation

Selection

Guidance → Safety mode → Confirmation → Confirm

Guidance → Safety mode → Confirmation → Confirm

• No
• Yes

"Confirmation" wizard
Navigation

Guidance → Safety mode → Confirmation

Output current transfer function
Navigation

Guidance → Safety mode → Confirmation → Curr. trans.func

Description

Assigned transfer function for current output.
NOTE"Square root" is only available with differential pressure measurement.

User interface

88• Linear• Square root
Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Low cutoff
Navigation

Guidance → Safety mode → Confirmation → Low cutoff

User interface

Character string comprising numbers, letters and special characters

ConfirmNavigation

Selection

• No
• Yes

Guidance → Safety mode → Confirmation → Confirm"Confirmation" wizard
Navigation

Guidance → Safety mode → Confirmation

Measuring mode current output
Navigation

Guidance → Safety mode → Confirmation → Output mode

Description

Assigned setting of curve form of current output.

User interface• Standard• Inverse• Bi-directional

Failure behavior current output
NavigationDescription

Assigned value of current output in case of an error.

User interface• Min.
• Max.

Endress+Hauser

Guidance → Safety mode → Confirmation → Failure behav.

89Description of device parameters

Cerabar PMP71B HART

Current range output
Navigation

Guidance → Safety mode → Confirmation → Current range

Description

Assigned current range used to transmit the measured value.

User interface• 4...20 mA (4...20.5 mA)• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• Customer specific

Lower range value output
Navigation

Guidance → Safety mode → Confirmation → Low.range outp

Description

Assigned value 4 mA.

User interface

Character string comprising numbers, letters and special characters

Upper range value output
Navigation

Guidance → Safety mode → Confirmation → Upp.range outp

Description

Assigned value 20 mA.

User interface

Character string comprising numbers, letters and special characters

Assign PV
NavigationDescription

Identifies the process variable linked with the primary variable. Primary variable is used in
HART as current output.

User interface• Pressure• Scaled variable

90

Guidance → Safety mode → Confirmation → Assign PV

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

ConfirmNavigation

Selection

• No
• Yes

Guidance → Safety mode → Confirmation → Confirm"Confirmation" wizard
Navigation

Guidance → Safety mode → Confirmation

Zero adjustment offset
Navigation

Guidance → Safety mode → Confirmation → Zero offset

Description

Assigned value of zero adjustment due to mounting position.

User interface

Character string comprising numbers, letters and special characters

ConfirmNavigation

Selection

• No
• Yes

Guidance → Safety mode → Confirmation → Confirm"Locking" wizard
Navigation

Guidance → Safety mode → Locking

Enter safety locking code

NavigationDescription

Enter the locking code to start the SIL/WHG locking sequence.

User entry

0 to 65 535

Endress+Hauser

Guidance → Safety mode → Locking → Safety code91Description of device parameters

Additional information

Cerabar PMP71B HART

Locking codes• WHG: 7450
• SIL: 7452
• SIL and WHG: 7454

Locking status
Navigation

Guidance → Safety mode → Locking → Locking status

Description

Displays the active write protection.

User interface• Hardware locked• Safety locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.
Selection
Function scope of the "Locking status" parameter
Options

Description

None

The access status displayed in the Access status display parameter applies. Only
appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the main electronics module.
This prevents write access to the parameters (e.g. via the local display or operatingtool).

Temporarily locked

Write access to the parameters is temporarily locked due to device-internal
processing (e.g. data upload/download, reset). Once the internal processing has
been completed, the parameters can be changed once again.

Save protocol?
NavigationDescription

The Report can be saved for archiving.

Selection

• No
• Yes

92

Guidance → Safety mode → Locking → Save protocol?

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Finish" wizard
Navigation

Guidance → Proof test → Finish

Inspector
Navigation

Guidance → Proof test → Finish → Inspector

Guidance → Safety mode → Result → Inspector

Description

The entered inspector name will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)

Location
Navigation

Guidance → Proof test → Finish → Location

Guidance → Safety mode → Result → Location

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)Date/time
Navigation

Guidance → Safety mode → Result → Date/time

User interface

Character string comprising numbers, letters and special characters

Notes
Navigation

Guidance → Proof test → Finish → Notes



Guidance → Safety mode → Result → Notes

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (255)

Endress+Hauser

93Description of device parameters

Cerabar PMP71B HART

Plant operator
Navigation

Guidance → Proof test → Finish → Plant operator



Guidance → Safety mode → Result → Plant operator

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)3.2.4

"Proof test" wizard

Navigation

Guidance → Proof test

"Preparation Proof test" wizard
Navigation

Guidance → Proof test → Prep. test

I have read the warning texts.
NavigationDescription

Loss of Safety function of protective system using safety-related output signal 4 to 20 mA- Read Functional safety manual or WHG documentation.
- Do not use safety-related output signal 4 to 20 mA for Safety function of protective
system- Suitable measures must be taken to guarantee process safety during the proof test

Selection

Yes

94

Guidance → Proof test → Prep. test → warning texts

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Preparation Proof test" wizard
Navigation

Guidance → Proof test → Prep. test

Visual inspection
Navigation

Guidance → Proof test → Prep. test → Visual inspect.

Description

The visual inspection may include:- Cable gland- Wiring- Terminal block- Housing/ housing lid- Mechanical and electrical installation
For further information refer to safety manual or WHG documentation.

Selection• Please select• Failed• Passed

Notes
Navigation

Guidance → Proof test → Prep. test → Notes

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (255)"Preparation Proof test" wizard
Navigation

Guidance → Proof test → Prep. test

Device tag
NavigationDescription

Displays a unique name for the measuring point so it can be identified quickly within theplant. The name is displayed in the header.

User interface

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Endress+Hauser

Guidance → Proof test → Prep. test → Device tag

95

Description of device parameters

Additional information

Cerabar PMP71B HART

User interface
1

XXXXXXXXX

A0029422

1

Position of the header text on the display

The number of characters displayed depends on the characters used.

Device name
Navigation

Guidance → Proof test → Prep. test → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

Serial number



Navigation

Guidance → Proof test → Prep. test → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

Firmware version
NavigationDescription

Displays the device firmware version that is installed.

User interface

Character string in the format xx.yy.zz

96

Guidance → Proof test → Prep. test → Firmware version

Endress+Hauser

Cerabar PMP71B HART

Additional information

Description of device parameters

User interface
The Firmware version is also located:• On the title page of the Operating instructions• On the transmitter nameplate

Hardware version
Navigation

Guidance → Proof test → Prep. test → Hardware version

User interface

Character string comprising numbers, letters and special characters"Preparation Proof test" wizard
Navigation

Guidance → Proof test → Prep. test

CRC device configuration
Navigation

Guidance → Proof test → Prep. test → CRC device conf.

Description

CRC device configuration based on current settings of safety relevant parameters.
The CRC device configuration is unique and can be used to detect changes in safety
relevant parameter settings.

User interface

0 to 65 535

Stored CRC device configuration
Navigation

Guidance → Proof test → Prep. test → Stored CRC conf.

Description

Stored CRC after the last safety lock. Factory delivery is 65535 means that the device has
not yet been safety locked.

User interface

0 to 65 535

Timestamp stored CRC device config.
NavigationDescription

Gives the time stamp when the CRC was last stored following completion of the safety lockwizard.

Endress+Hauser

Guidance → Proof test → Prep. test → Time stored CRC

97

Description of device parameters

User interface

Cerabar PMP71B HART

Character string comprising numbers, letters and special characters

Operating time
Navigation

Guidance → Proof test → Prep. test → Operating time

Description

Indicates how long the device has been in operation.

Additional information

Maximum time: 9 999 d (≈ 27 years)

Configuration counter
Navigation

Guidance → Proof test → Prep. test → Config. counter

Description

Displays the counter for changes to the device parameters.
Additional information:- If the value for a static parameter is changed when optimizing or configuring theparameter, the counter is incremented by 1. This is to enable tracking different parameterversions.
- When multiple parameters are changed simultaneously, e.g. when loading parameters
into the device from an external source such as FieldCare, the counter may display a higher
value. The counter cannot be reset, nor is it reset to a default value on performing a device
reset.
- Once the counter has reached the value 65535, it restarts at 0.

User interface

98

0 to 65 535

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Preparation Proof test" wizard
Navigation

Guidance → Proof test → Prep. test

Select test procedure
Navigation

Guidance → Proof test → Prep. test → Select test proc

Description

The proof test of the device can be done using the following test procedures:
Test procedure A with wizard (PTC > 90%):- Simulate and check the min and max alarm current.
- Approach and check the lower and upper measured value.
- Confirm safety function.
Test procedure B with wizard (PTC > 50%):-Verify the current measured value with a plausibility check.
- Simulate and check the min and max alarm current.
- Confirm safety function.
For additional test procedures without wizard, refer to safety manual.

Selection• Please select• Test procedure A• Test procedure B

Safety function?
Navigation

Guidance → Proof test → Prep. test → Safety function?

Selection• Please select• MAX monitoring• MIN monitoring• Range monitoring"Simulation and check failure current" wizard
Navigation

Guidance → Proof test → Simulation

Failure current (high alarm)
NavigationDescription

Actually set value for high failure current.

Endress+Hauser

Guidance → Proof test → Simulation → Fail. current (high alarm)99Description of device parameters

User interface

Cerabar PMP71B HART

21.5 to 23 mA

Confirm
Navigation

Guidance → Proof test → Simulation → Confirm

Description

Confirm that the displayed value corresponds to the set high failure current.

Selection• Please select
• No
• Yes

(High) Is safety function triggered?
Navigation

Guidance → Proof test → Simulation → (High)Funct. trigg?

Description

Confirm that the output current triggers the required safety function.

Selection

Yes

Terminal current (high alarm)
Navigation

Guidance → Proof test → Simulation → Term. curr. (HA)

Description

Internally readback terminal current (high alarm).

User interface

Signed floating-point number

Confirm
NavigationDescription

Confirm that the readback terminal current corresponds to the set high failure current
within tolerance +-0,32mA.

Selection• Please select
• No
• Yes

100

Guidance → Proof test → Simulation → Confirm

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters(Low) Is safety function triggered?
Navigation

Guidance → Proof test → Simulation → (Low)Funct. trigg?

Description

Confirm that the output current triggers the required safety function.

Selection

Yes"Proof test" wizard
Navigation

Guidance → Proof test → Proof test

Setpoint 1
Navigation

Guidance → Proof test → Proof test → Setpoint 1

Description

Physically apply a reference pressure for upper range.

User entry

Signed floating-point number

Pressure 1
Navigation

Guidance → Proof test → Proof test → Pressure 1

Description

Pressure output

User interface

Signed floating-point number

Terminal current 1
NavigationDescription

Terminal current for upper range.

User interface

0 to 30 mA

Endress+Hauser

Guidance → Proof test → Proof test → Terminal curr. 1

101

Description of device parameters

Cerabar PMP71B HART

Current deviation 1
Navigation

Guidance → Proof test → Proof test → Curr. deviati 1

Description

Decide if the terminal current for upper range corresponding to the applied pressure is
within tolerance.

Selection• Please select• Failed• Passed

Setpoint 2
Navigation

Guidance → Proof test → Proof test → Setpoint 2

Description

Physically apply a reference pressure for lower range.

User entry

Signed floating-point number

Pressure 2
Navigation

Guidance → Proof test → Proof test → Pressure 2

Description

Pressure output

User interface

Signed floating-point number

Terminal current 2
Navigation

Guidance → Proof test → Proof test → Terminal curr. 2

Description

Terminal current for lower range.

User interface

0 to 30 mA

Current deviation 2
NavigationDescription

Decide if the terminal current for lower range corresponding to the applied pressure is
within tolerance.

102

Guidance → Proof test → Proof test → Curr. deviati 2

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters• Please select• Failed• Passed"Finish" wizard
Navigation

Guidance → Proof test → Finish

Active diagnostics
Navigation

Guidance → Proof test → Finish → Active diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the current diagnostic message. If two or more messages occur simultaneously,
the message with the highest priority is shown on the display.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Additional pending diagnostic messages can be viewed in the Diagnostic list
submenu.
ExampleFor the display format:
F271 Main electronic failure

Summary
Navigation

Guidance → Proof test → Finish → Summary

User interface• Passed• Failed• Unknown

Decision of inspector
NavigationDescription

Responsible for the evaluation of the proof test is the inspector.
The summary displayed is not a binding inspection decision.

Endress+Hauser

Guidance → Proof test → Finish → Decision of insp

103

Description of device parameters

Selection

Cerabar PMP71B HART

• Please select• Failed• Passed

Inspector
Navigation

Guidance → Proof test → Finish → Inspector

Description

The entered inspector name will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)

Location
Navigation

Guidance → Proof test → Finish → Location

Description

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)Date/time proof test
Navigation

Guidance → Proof test → Finish → Date/time

Description

This value is updated with every proof test and with inspector decision „Passed“.

User interface

Character string comprising numbers, letters and special characters

Notes
NavigationDescription

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (255)

104

Guidance → Proof test → Finish → Notes

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Plant operator
NavigationDescription

The entered value will be included in the report.

User entry

Character string comprising numbers, letters and special characters (96)Endress+Hauser

Guidance → Proof test → Finish → Plant operator

105

Description of device parameters

Cerabar PMP71B HART

3.3

"Diagnostics" menu

Navigation

3.3.1



Diagnostics"Active diagnostics" submenu

Navigation

Diagnostics → Active diagnos.

Active diagnostics
Navigation

 Diagnostics → Active diagnos. → Active diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the current diagnostic message. If two or more messages occur simultaneously,
the message with the highest priority is shown on the display.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Additional pending diagnostic messages can be viewed in the Diagnostic list
submenu.
ExampleFor the display format:
F271 Main electronic failure

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the operating time when the current diagnostic message occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be viewed via the Actual diagnostics parameter(→  103).
ExampleFor the display format:
24d12h13m00s

106Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Previous diagnostics
Navigation

 Diagnostics → Active diagnos. → Prev.diagnostics

Prerequisite

Two diagnostic events have already occurred.

Description

Displays the diagnostic message that occurred before the current message.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
ExampleFor the display format:
F271 Main electronic failure

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the operating time when the last diagnostic message before the current message
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be viewed via the Previous diagnostics parameter(→  107).
ExampleFor the display format:
24d12h13m00s

Operating time from restart
Navigation

 Diagnostics → Active diagnos. → Time fr. restart

Description

Indicates how long the device has been in operation since the last time the device wasrestarted.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

107Description of device parameters

Cerabar PMP71B HART

Operating time
Navigation

 Diagnostics → Active diagnos. → Operating time

Description

Indicates how long the device has been in operation.

Additional information

Maximum time: 9 999 d (≈ 27 years)

3.3.2"Event logbook" submenu

Navigation

Diagnostics → Event logbook

Filter options
Navigation

Diagnostics → Event logbook → Filter options

Description

Use this function to select the category whose event messages are displayed in the event
list of the operating tool.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)• Not categorized

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required

Clear event listNavigationDescription

Use this function to process the current values in the event logbook.

Selection• Cancel• Clear data

108

Diagnostics → Event logbook → Clear event list

Endress+Hauser

Cerabar PMP71B HART

Additional information

Description of device parameters

Description
Once this function has been executed, the events list is empty and all the events aredeleted.
The events list can be exported using an operating tool (e.g. FieldCare.

3.3.3"Minimum/maximum values" submenu

Navigation

Diagnostics → Min/max val.

Pressure min
Navigation

 Diagnostics → Min/max val. → Pressure min

Description

Minimum or maximum value measured by device.

User interface

Signed floating-point number

Counter limit underruns sensor Pmin
Navigation

 Diagnostics → Min/max val. → Counter P < Pmin

Description

Counts how many times the value underruns the sensor specific minimum values.
Sensor specific minimum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter underruns of user limit Pmin
Navigation

 Diagnostics → Min/max val. → Counter < P user

Description

Counts how many times the value underruns the minimum values defined by the user.
User defined minimum values are shown in Diagnostic/Diagnostic settings/Properties
menu.

User interface

0 to 65 535

Endress+Hauser

109

Description of device parameters

Cerabar PMP71B HART

Minimum sensor temperature
Navigation

 Diagnostics → Min/max val. → Min. sensor temp

Description

Minimum or maximum value measured by device.
Users cannot reset this value.

User interface–273.15 to 9 726.85 °C

Counter limit underruns sensor Tmin
Navigation

 Diagnostics → Min/max val. → Counter T < Tmin

Description

Counts how many times the value underruns/overruns the sensor specific minimum/
maximum values.
Sensor specific minimum/maximum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter underruns of user limit Tmin
Navigation

 Diagnostics → Min/max val. → Counter < T user

User interface

0 to 65 535

Minimum terminal voltage
Navigation

 Diagnostics → Min/max val. → Min.term.volt.

Description

Minimum or maximum measured terminal (supply) voltage.

User interface

0.0 to 50.0 V

Minimum electronics temperature
Navigation

 Diagnostics → Min/max val. → Min.electr.temp.

Description

Minimum or maximum measured main electronics temperature.

User interface

Signed floating-point number

110

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Reset user defined counters P and T
Navigation

 Diagnostics → Min/max val. → Reset count. P T

Selection• Cancel• ConfirmPressure max
Navigation

 Diagnostics → Min/max val. → Pressure max

Description

Minimum or maximum value measured by device.

User interface

Signed floating-point number

Counter limit overruns sensor Pmax
Navigation

 Diagnostics → Min/max val. → Counter P > Pmax

Description

Counts how many times the value overruns the sensor specific maximum values.
Sensor specific maximum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter overruns of user limit Pmax
Navigation

 Diagnostics → Min/max val. → Counter > P user

Description

Counts how many times the value overruns the maximum values defined by the user.
User defined maximum values are shown in Diagnostic/Diagnostic settings/Properties
menu.

User interface

0 to 65 535

Maximum sensor temperature
Navigation

 Diagnostics → Min/max val. → Max. sensor temp

Description

Minimum or maximum value measured by device.
Users cannot reset this value.

User interface–273.15 to 9 726.85 °C

Endress+Hauser

111Description of device parameters

Cerabar PMP71B HARTCounter limit overruns sensor Tmax
Navigation

 Diagnostics → Min/max val. → Counter T > Tmax

Description

Counts how many times the value underruns/overruns the sensor specific minimum/
maximum values.
Sensor specific minimum/maximum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter overruns of user limit Tmax
Navigation

 Diagnostics → Min/max val. → Counter > T user

User interface

0 to 65 535

Maximum terminal voltage
Navigation

 Diagnostics → Min/max val. → Max.term.voltage

Description

Minimum or maximum measured terminal (supply) voltage.

User interface

0.0 to 50.0 V

Maximum electronics temperature
Navigation

 Diagnostics → Min/max val. → Max.electr.temp.

Description

Minimum or maximum measured main electronics temperature.

User interface

Signed floating-point number

112

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3.3.4"Simulation" submenu

Navigation

Diagnostics → Simulation

Simulation



Navigation

 Diagnostics → Simulation → Simulation

Description

Simulates one or more process variables and/or events.
Warning:
Output will reflect the simulated value or event.

Selection• Off• Current output• Diagnostic event simulation• Pressure

Value pressure simulation



Navigation

 Diagnostics → Simulation → Pressure

User entry

Signed floating-point number

Value current output

Navigation

 Diagnostics → Simulation → Current output

Description

Defines the value of the simulated output current.

User entry

3.59 to 23 mA

Diagnostic event simulation



Navigation

 Diagnostics → Simulation → Diagnostic event

Description

Use this function to select a diagnostic event for the simulation process that is activated.

Selection• Off• Diagnostic event picklist (depends on the category selected)

Endress+Hauser

113

Description of device parameters

Additional information

Cerabar PMP71B HART

Description
For the simulation, you can choose from the diagnostic events of the category selectedin the Diagnostic event category parameter.

3.3.5
Navigation"Heartbeat Technology" submenu
Diagnostics → Heartbeat Techn.

"Heartbeat Verification" submenu
Navigation

Diagnostics → Heartbeat Techn. → Heartbeat Verif.

Date/time Heartbeat Verification
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Date/time Heartbeat
Verification

Description

Date and time of last Heartbeat Verification.
This value is updated with every Heartbeat verification.
Note:
If time information is not available, e.g. Heartbeat verification is started from display,'--------' is shown.

User interface

Character string comprising numbers, letters and special characters

Operating time (Verification)
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Operating time

User interface

Days (d), hours (h), minutes (m), seconds (s)Verification result
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Verific. result

User interface• Not done
• Passed• Not done
• Failed

114

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

StatusNavigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Status

Description

Displays the current status of the verification.

User interface• Done• Busy• Failed• Not done

"Loop diagnostics" submenu
Navigation

Diagnostics → Heartbeat Techn. → Loop diagn.

Rebuild baselineNavigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Reb. baseline

Description

Notice
The current output is simulated.
Bridge the PLC or take other appropriate measures to prevent an erroneous triggering ofalarm messages or changes in the control loop behavior.
The baseline should be rebuilt if planned changes have been made in the loop.

Selection

• No
• Yes

Tolerated deviation +/-Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Toler. deviation

Description

A value should be chosen to ensure that normal voltage deviations do not lead tounwanted messages.
Default
1.5 V DC

User entry

Endress+Hauser

0.5 to 3.0 V

115

Description of device parameters

Cerabar PMP71B HART

Baseline status
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Baseline status

Description"Failed"Means, baseline is not available or creation not possible.
"Success"
Baseline is available.

User interface• Failed• Success

Loop diagnostics



Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Loop diagn.

Description

Enable/disable loop diagnostics.
Note:
If the function is disabled, there is no analysis and no event message.

Selection• Disable• Enable

Terminal voltage 1
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Terminal volt. 1

Description

Shows the current terminal voltage that is applied at the output

User interface

0.0 to 50.0 V

Clamping voltage lower threshold
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Lower threshold

User interface

0.0 to 50.0 V

Clamping voltage upper threshold
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Upper threshold

User interface

0.0 to 50.0 V

116

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

806 Event delay

Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → 806 Event delay

Description

Displays how long the triggering status must be present until an event message is issued.
Used to filter out short-term signal interference.

User entry

0 to 60 s"Statistical Sensor Diagnostics" submenu
Navigation

Diagnostics → Heartbeat Techn. → SSD

SSD: Statistical Sensor Diagnostics
Navigation

 Diagnostics → Heartbeat Techn. → SSD → Stat. Sens. Diag

Description

Enable or disable SSD.



After selecting "Disable", no statistical sensor diagnosis takes place. No diagnostic messages
are output.
Selection• Disable• Enable

System status
Navigation

 Diagnostics → Heartbeat Techn. → SSD → System status

User interface• Idle• No sufficient signal noise• Stable• Not stable• Verify System Dynamics• Process dynamic too highSignal status
Navigation

 Diagnostics → Heartbeat Techn. → SSD → Signal status

User interface• Idle• Building Baseline• Verifying Baseline

Endress+Hauser

117Description of device parameters

Cerabar PMP71B HART

• Verifying baseline failed• Monitoring• Out of range• Monitoring inactive

Signal noise status
Navigation

 Diagnostics → Heartbeat Techn. → SSD → Noise status

User interface• Idle• Building Baseline• Verifying Baseline• Verifying baseline failed• Monitoring• Out of range• Monitoring inactive

Counter Baseline creation SSD
Navigation

 Diagnostics → Heartbeat Techn. → SSD → Counter Baseline

Description

Specifies how often the baseline has been rebuilt.

User interface

Positive integer

3.3.6"Diagnostic settings" submenu

Navigation

Diagnostics → Diag. settings"Properties" submenu
Navigation

Diagnostics → Diag. settings → Properties

SSD Out of range delay timeNavigation

User entry

0 to 604 800 s

118

Diagnostics → Diag. settings → Properties → SSD Delay time

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

SSD Monitoring delay timeNavigation

Diagnostics → Diag. settings → Properties → SSD Verz. Zeit

User entry

0 to 86 400 s

500 Process alert pressureNavigation

Diagnostics → Diag. settings → Properties → 500 Pressure

Description

Define whether user-defined pressure limits should be set.
If "Off" is selected, no analysis will take place and no event message will be generated.

Selection

• Off
• On

Low alert value

Navigation

Diagnostics → Diag. settings → Properties → Low alert value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

High alert value

Navigation

Diagnostics → Diag. settings → Properties → High alert value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

501 Process alert scaled variable

NavigationDescription

Define whether user-defined limits should be set.
If "Off" is selected, no analysis will take place and no event message will be generated.

Endress+Hauser

Diagnostics → Diag. settings → Properties → 501 Scaled var.

119

Description of device parameters

Selection

Cerabar PMP71B HART

• Off
• On

Low alert value

Navigation

Diagnostics → Diag. settings → Properties → Low alert value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

High alert value

Navigation

Diagnostics → Diag. settings → Properties → High alert value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry

Signed floating-point number

User temperature process alert

Navigation

Diagnostics → Diag. settings → Properties → UserTemp alert

Description

Define whether the user-defined sensor temperature limits should be set. If "Off" noanalysis and therefore no event message will take place.

Selection

• Off
• On

Low alert value

NavigationDescription

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry–50 to 150 °C

120

Diagnostics → Diag. settings → Properties → Low alert value

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

High alert value

Navigation

Diagnostics → Diag. settings → Properties → High alert value

Description

Set range.
If this limit value is exceeded or undercut, a diagnostic event is generated. There is no
hysteresis.

User entry–50 to 150 °C

806 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Properties → 806 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

806 Event category



Navigation

 Diagnostics → Diag. settings → Properties → 806Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)806 Event delay

NavigationDescription

Displays how long the triggering status must be present until an event message is issued.
Used to filter out short-term signal interference.

User entry

0 to 60 s

Endress+Hauser

Diagnostics → Diag. settings → Properties → 806 Event delay

121

Description of device parameters

Cerabar PMP71B HART

"Configuration" submenu
Navigation

Diagnostics → Diag. settings → Configuration → Configuration

500 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 500 Diag. behav.

Description

Select event behavior"Logbook entry only":
no digital or analog transmission of the message"Warning": Current output unchanged. Message is output digitally (default).
"Alarm": Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

500 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 500Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)501 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 501 Diag. behav.

Description

Select event behavior"Logbook entry only":
no digital or analog transmission of the message"Warning": Current output unchanged. Message is output digitally (default).
"Alarm": Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

122

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters• Off• Alarm• Warning• Logbook entry only

501 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 501Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)502 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 502 Diag. behav.

Description

Select event behavior"Logbook entry only":
no digital or analog transmission of the message"Warning": Current output unchanged. Message is output digitally (default).
"Alarm": Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

502 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 502Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Endress+Hauser

123Description of device parameters

Cerabar PMP71B HART

"Process" submenu
Navigation

Diagnostics → Diag. settings → Configuration → Process

806 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 806 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

806 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 806Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)822 Diagnostic behavior
Navigation

User interface• Alarm• Warning• Logbook entry only

124

Diagnostics → Diag. settings → Configuration → Process → 822 Diag. behav.

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

822 Event category



Navigation

Diagnostics → Diag. settings → Configuration → Process → 822 Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Sensor pressure range behavior

Navigation

Diagnostics → Diag. settings → Configuration → Process → P-range behavior

Description

Select event behavior"Alarm":Current output adopts the set alarm current.
"Warning":
Current output unchanged. Message is displayed digitally (factory setting).
"Logbook entry only":
No digital or analog forwarding of the message.
"Special":– Lower sensor limit undercut: Current output < 3.6 mA.
– Upper sensor limit exceeded: Current output 21 to 23 mA, depending on the setting.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning message disappears.

Selection• Alarm• Warning• Logbook entry only
• Special

841 Event category



Navigation

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Endress+Hauser

Diagnostics → Diag. settings → Configuration → Process → 841 Event category

125Description of device parameters

Cerabar PMP71B HART

900 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 900Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)900 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 900 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

906 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 906 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Off• Warning• Logbook entry only

906 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 906Event category

Description

Select category for diagnostic message.

126

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)"Process" submenu
Navigation

Diagnostics → Diag. settings → Configuration → Process

806 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 806 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

806 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 806Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)822 Diagnostic behavior
Navigation

User interface• Alarm• Warning• Logbook entry only

Endress+Hauser

Diagnostics → Diag. settings → Configuration → Process → 822 Diag. behav.

127

Description of device parameters

Cerabar PMP71B HART

822 Event category



Navigation

Diagnostics → Diag. settings → Configuration → Process → 822 Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Sensor pressure range behavior

Navigation

Diagnostics → Diag. settings → Configuration → Process → P-range behavior

Description

Select event behavior"Alarm":Current output adopts the set alarm current.
"Warning":
Current output unchanged. Message is displayed digitally (factory setting).
"Logbook entry only":
No digital or analog forwarding of the message.
"Special":– Lower sensor limit undercut: Current output < 3.6 mA.
– Upper sensor limit exceeded: Current output 21 to 23 mA, depending on the setting.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning message disappears.

Selection• Alarm• Warning• Logbook entry only
• Special

841 Event category



Navigation

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)128

Diagnostics → Diag. settings → Configuration → Process → 841 Event category

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

900 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 900Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)900 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 900 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

906 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 906 Diag. behav.

Description

Select event behavior"Logbook entry only": no digital or analog transmission of the message.
"Warning": Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Off• Warning• Logbook entry only

906 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 906Event category

Description

Select category for diagnostic message.

Endress+Hauser

129

Description of device parameters

Selection

130

Cerabar PMP71B HART

• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3.4"Application" menu

Navigation

3.4.1



Application"Measuring units" submenu

Navigation

Application → Measuring units

Pressure unitNavigation

 Application → Measuring units → Pressure unit

Selection

SI units
• MPa
• kPa
• Pa
• bar
• mbar• torr• atm• kgf/cm²• gf/cm²

US unitspsi

Other units• inH2O• inH2O (4°C)• mmH2O• mmH2O (4°C)• mH2O• mH2O (4°C)• ftH2O• inHg• mmHg

Decimal places pressureNavigation

Application → Measuring units → Decimal pressure

Description

This selection does not affect the measurement and calculation accuracy of the device.

Selection• Automatic
The decimal place is configured automatically.
Example: Unit mbar: one decimal place. Unit bar: four decimal places•x• x.x• x.xx• x.xxx• x.xxxx

Temperature unitNavigation

 Application → Measuring units → Temperature unit

Description

Use this function to select the unit for the temperature.

Endress+Hauser

131Description of device parameters

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

Selection

Cerabar PMP71B HART

US units°F

Scaled variable unitNavigation

 Application → Measuring units → Scaled Unit

Description

Use "Free text", first selection, if the desired unit is not available in the selection list. It is
possible to define a customer specific unit with another parameter.

132

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters

SI units
•%
• mm
• cm
•m
•l
• hl
• m³
•g
• kg
•t
• g/s
• kg/s• kg/min• kg/h• t/min• t/h• t/d• m³/s• m³/min• m³/h• m³/d• l/s• l/min• l/h• Nm³/h• Nl/h• Sm³/s• Sm³/min• Sm³/h• Sm³/d• Nm³/s• g/cm³• kg/m³• Nm³/min• Nm³/dUS units
• ft
• in
• ft³
• gal (us)• bbl (us;oil)
• oz
• lb
• STon• lb/s• lb/min• lb/h• STon/min• STon/h• STon/d• ft³/s• ft³/min• ft³/h• ft³/d• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• Sft³/min• Sft³/h• Sft³/d

Imperial units• gal (imp)• gal/s (imp)• gal/min (imp)• gal/h (imp)

Custom-specific units
Free text

Free text

Navigation

 Application → Measuring units → Free text

User entry

Character string comprising numbers, letters and special characters (32)

Decimal places scaled variable

NavigationDescription

This selection does not affect the measurement and calculation accuracy of the device.

Endress+Hauser

Application → Measuring units → Decimal scaled133Description of device parameters

Selection

Cerabar PMP71B HART

•x
• x.x
• x.xx• x.xxx• x.xxxx

3.4.2
Navigation"Measured values" submenu
Application → Measured values

Sensor pressure
Navigation

 Application → Measured values → Sensor pressure

User interface

Signed floating-point number

Pressure
Navigation

 Application → Measured values → Pressure

Scaled variable
Navigation

 Application → Measured values → Scaled variable

User interface

Signed floating-point number

Sensor temperature
Navigation

 Application → Measured values → Sensor temp.

User interface–273.15 to 9 726.85 °C

Terminal voltage 1
Navigation

 Application → Measured values → Terminal volt. 1

Description

Shows the current terminal voltage that is applied at the output

134

Endress+Hauser

Cerabar PMP71B HART

User interface

Description of device parameters

0.0 to 50.0 V

Terminal current
Navigation

 Application → Measured values → Terminal curr.

Description

Shows the current value of the current output which is currently measured

User interface

0 to 30 mA

Electronics temperature
Navigation

 Application → Measured values → Electronics temp

Description

Displays the current temperature of the main electronics.

User interface

Signed floating-point number

3.4.3
Navigation"Sensor" submenu
Application → Sensor"Basic settings" submenu
Navigation

Application → Sensor → Basic settings

Output current transfer function
Navigation

 Application → Sensor → Basic settings → Curr. trans.func

Description

Linear
The linear pressure signal is used for the current output. The flow must be calculated in
the evaluation unit.
Square root - differential pressure only
The root flow signal is used for the current output. The 'Flow (square root)'current signal is indicated on the on-site display with a root symbol.

Endress+Hauser

135

Description of device parameters

User interface

Cerabar PMP71B HART

• Linear• Square root *DampingNavigation

 Application → Sensor → Basic settings → Damping

Description

The damping is effective before the measured value is further processed, i.e., before thefollowing processes:- Scaling- Limit value monitoring- Forwarding to display- Forwarding to Analog Input BlockNote:
The Analog Input Block has its own “Damping” parameter. In the measurement chain, only
one of the two attenuation parameters shall have a value other than 0.
Otherwise, the signal will be attenuated several times.

User entry

0 to 999.0 s"Sensor calibration" submenu
Navigation

Application → Sensor → Sensor cal.

Zero adjustment

Navigation

 Application → Sensor → Sensor cal. → Zero adjustment

Description

Due to the mounting position of the measuring instrument, a pressure shift may occur.
The pressure shift can be corrected with the zero adjustment.

Selection• No• Confirm

Calibration offset

Navigation

 Application → Sensor → Sensor cal. → Calibr offset

Prerequisite

Absolute pressure sensor*

136

Visibility depends on order options or device settings

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Description

Enter the value by which the measured value should be corrected, e.g., a position
adjustment for absolute pressure sensors.

User entry

Signed floating-point number

Zero adjustment offset

Navigation

 Application → Sensor → Sensor cal. → Zero offset

User entry

Signed floating-point number

Sensor Trim Reset

Navigation

Application → Sensor → Sensor cal. → Sen. Trim Reset

Selection• No• Confirm

Lower sensor trim measured value
Navigation

User interface

Signed floating-point number

Endress+Hauser

Application → Sensor → Sensor cal. → LowerTrimMeasVal

137

Description of device parameters

Cerabar PMP71B HART

Lower sensor trim

Navigation

Application → Sensor → Sensor cal. → LowerSensor trim

Description

These two parameters allow a recalibration of the sensor, i.e., if you want to fit the sensor
to the measuring range. The highest accuracy is obtained when the value for the "Lower
sensor trim" is as close as possible to "LRV" (lower range value) and the value for "Upper
sensor trim" as close as possible to "URV" (upper range value).
There must be a known reference pressure when setting a new lower or upper sensor
characteristic curve value. The more accurate the reference pressure is during
recalibration, the higher the accuracy of the pressure transmitter later. A new value isassigned to the applied pressure using "Lower sensor trim" and "Upper sensor trim"
parameters.
Note:
The value entered can be at maximum "Sensor pressure" +/- 10 % of the permitted
maximum pressure (URL).
Proceed as follows:- Apply reference pressure for lower range value ("LRV")- Enter the measured reference pressure at "Lower sensor trim" and confirm- Apply reference pressure for upper range value ("URV")- Enter the measured reference pressure at "Upper sensor trim" and confirm- The sensor is now calibrated

User entry

Signed floating-point number

Upper sensor trim measured value
Navigation

User interface

Signed floating-point number

138

Application → Sensor → Sensor cal. → UpperTrimMeasVal

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Upper sensor trim

Navigation

Application → Sensor → Sensor cal. → UpperSensor trim

Description

These two parameters allow a recalibration of the sensor, i.e., if you want to fit the sensor
to the measuring range. The highest accuracy is obtained when the value for the "Lower
sensor trim" is as close as possible to "LRV" (lower range value) and the value for "Upper
sensor trim" as close as possible to "URV" (upper range value).
There must be a known reference pressure when setting a new lower or upper sensor
characteristic curve value. The more accurate the reference pressure is during
recalibration, the higher the accuracy of the pressure transmitter later. A new value isassigned to the applied pressure using "Lower sensor trim" and "Upper sensor trim"
parameters.
Note:
The value entered can be at maximum "Sensor pressure" +/- 10 % of the permitted
maximum pressure (URL).
Proceed as follows:- Apply reference pressure for lower range value ("LRV")- Enter the measured reference pressure at "Lower sensor trim" and confirm- Apply reference pressure for upper range value ("URV")- Enter the measured reference pressure at "Upper sensor trim" and confirm- The sensor is now calibrated

User entry

Signed floating-point number"Sensor limits" submenu
Navigation

Application → Sensor → Sensor limits

Lower Range Limit
Navigation

Application → Sensor → Sensor limits → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
NavigationDescription

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

Endress+Hauser

Application → Sensor → Sensor limits → URL

139

Description of device parameters

Cerabar PMP71B HART

Minimum span
Navigation

Application → Sensor → Sensor limits → Minimum span

Description

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number

Sensor temperature lower range limit
Navigation

Application → Sensor → Sensor limits → Sens.temp.lo.lim

User interface–273.15 to 9 726.85 °C

Sensor temperature upper range limit
Navigation

Application → Sensor → Sensor limits → Sens.temp.up.lim

User interface–273.15 to 9 726.85 °C"Scaled variable" submenu
Navigation

Application → Sensor → Scaled variable

Assign PV

Navigation

 Application → Sensor → Scaled variable → Assign PV

Description

Use this function to select a measured variable (HART device variable) for the primary
dynamic variable (PV).

Selection• Pressure• Scaled variable

140

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Scaled variable unitNavigation

 Application → Sensor → Scaled variable → Scaled Unit

Description

Use "Free text", first selection, if the desired unit is not available in the selection list. It is
possible to define a customer specific unit with another parameter.

Selection

SI units
•%
• mm
• cm
•m
•l
• hl
• m³
•g
• kg
•t
• g/s
• kg/s• kg/min• kg/h• t/min• t/h• t/d• m³/s• m³/min• m³/h• m³/d• l/s• l/min• l/h• Nm³/h• Nl/h• Sm³/s• Sm³/min• Sm³/h• Sm³/d• Nm³/s• g/cm³• kg/m³• Nm³/min• Nm³/dUS units
• ft
• in
• ft³
• gal (us)• bbl (us;oil)
• oz
• lb
• STon• lb/s• lb/min• lb/h• STon/min• STon/h• STon/d• ft³/s• ft³/min• ft³/h• ft³/d• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• Sft³/min• Sft³/h• Sft³/d

Imperial units• gal (imp)• gal/s (imp)• gal/min (imp)• gal/h (imp)

Custom-specific units
Free text

Free text

Navigation

 Application → Sensor → Scaled variable → Free text

User entry

Character string comprising numbers, letters and special characters (32)Endress+Hauser

141

Description of device parameters

Cerabar PMP71B HART

Pressure
Navigation

 Application → Sensor → Scaled variable → Pressure

Scaled variable transfer function



Navigation

 Application → Sensor → Scaled variable → Scaled function

Description"Linear'
The linear pressure signal is used for the output signal. The flow must be calculated in theevaluation unit.
"Square root" (Deltabar)
The root flow signal is used for the output signal. The "Flow (square root)'
output signal is indicated on the on-site display with a root symbol.
"Table'The output is defined according to the scaled variable / pressure table entered.

Selection• Linear• Square root *• TableLower range value output

Navigation

 Application → Sensor → Scaled variable → Low.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Upper range value output

Navigation

 Application → Sensor → Scaled variable → Upp.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number*

142

Visibility depends on order options or device settings

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Activate table

Navigation

Application → Sensor → Scaled variable → Activate table

Selection• Disable• EnablePressure value 1Navigation

 Application → Sensor → Scaled variable → Pressure 1

Description

Enter pressure for the first scaling point. "Scaled variable value 1" will be allocated to thispressure.

User entry

Signed floating-point number

Scaled variable value 1
Navigation

 Application → Sensor → Scaled variable → Scaled 1

Description

Enter value for the first scaling point. This value is allocated to "Pressure value 1".

User interface

Signed floating-point number

Pressure value 2Navigation

 Application → Sensor → Scaled variable → Pressure 2

Description

Enter pressure for the second scaling point. "Scaled variable value 2" will be allocated tothis pressure.

User entry

Signed floating-point number

Scaled variable value 2Navigation

 Application → Sensor → Scaled variable → Scaled 2

Description

Enter value for the second scaling point. This value is allocated to "Pressure value 2".

User entry

Signed floating-point number

Endress+Hauser

143

Description of device parameters

Cerabar PMP71B HART

Pressure

Navigation

Application → Sensor → Scaled variable → Pressure

User entry

Signed floating-point number

Scaled variable

Navigation

Application → Sensor → Scaled variable → Scaled variable

User entry

Signed floating-point number"Wet calibration" submenu
Navigation

Application → Sensor → Wet calibration

Zero



Navigation

 Application → Sensor → Wet calibration → Zero

Selection• No• Confirm

Pressure value 1Navigation

Application → Sensor → Wet calibration → Pressure 1

Description

Enter pressure for the first scaling point. "Scaled variable value 1" will be allocated to thispressure.

User entry

Signed floating-point number

Span



Navigation

 Application → Sensor → Wet calibration → Span

Selection• No• Confirm

144

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Pressure value 2Navigation

Application → Sensor → Wet calibration → Pressure 2

Description

Enter pressure for the second scaling point. "Scaled variable value 2" will be allocated tothis pressure.

User entry

Signed floating-point number

Lower range value output

Navigation

Application → Sensor → Wet calibration → Low.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Upper range value output

Navigation

Application → Sensor → Wet calibration → Upp.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number"Wet calibration" submenu
Navigation

Application → Sensor → Wet calibration

Zero



Navigation

 Application → Sensor → Wet calibration → Zero

Selection• No• Confirm

Endress+Hauser

145

Description of device parameters

Cerabar PMP71B HART

Pressure value 1Navigation

Application → Sensor → Wet calibration → Pressure 1

Description

Enter pressure for the first scaling point. "Scaled variable value 1" will be allocated to thispressure.

User entry

Signed floating-point number

Span



Navigation

 Application → Sensor → Wet calibration → Span

Selection• No• Confirm

Pressure value 2Navigation

Application → Sensor → Wet calibration → Pressure 2

Description

Enter pressure for the second scaling point. "Scaled variable value 2" will be allocated tothis pressure.

User entry

Signed floating-point number

Lower range value output

Navigation

Application → Sensor → Wet calibration → Low.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Upper range value output

NavigationDescription

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

146

Application → Sensor → Wet calibration → Upp.range outp

Endress+Hauser

Cerabar PMP71B HART

User entry

Description of device parameters

Signed floating-point number

3.4.4"Current output" submenu

Navigation

Application → Curr.output

Assign PV

Navigation

Application → Curr.output → Assign PV

Description

Use this function to select a measured variable (HART device variable) for the primary
dynamic variable (PV).

Selection• Pressure• Scaled variable

Measuring mode current output
Navigation

 Application → Curr.output → Output mode

Description

Select curve of current output.

Selection• Standard• Inverse• Bi-directional

Current range output



Navigation

 Application → Curr.output → Current range

Description

Defines the current range used to transmit the measured or calculated value.
In brackets are indicated the “low saturation value” and the “high saturation value”.
If Measured value ≤ “low saturation”, the output current is set to “low saturation”.
If Measured value ≥ “high saturation”, the output current is set to ”high saturation”.
Note:
Currents below 3.6 mA or above 21.5 mA can be used to signal an alarm.

Selection

Endress+Hauser• 4...20 mA (4...20.5 mA)• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)

147

Description of device parameters

Cerabar PMP71B HART

Lower range value output

Navigation

 Application → Curr.output → Low.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Upper range value output

Navigation

 Application → Curr.output → Upp.range outp

Description

Depending on which variable has been selected as "Process variable output current ", define
the related lower (4 mA) and upper range values (20 mA).

User entry

Signed floating-point number

Failure behavior current output

Navigation

 Application → Curr.output → Failure behav.

Description

Defines which current the output assumes in the case of an error.
Min: < 3.6 mA
Max: >21.5 mA
Note: The hardware DIP Switch for alarm current has priority over software setting.

Selection• Min.
• Max.

Failure current

NavigationDescription

Enter current output value in alarm condition

User entry

21.5 to 23 mA

148

Application → Curr.output → Failure current

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Output current
Navigation

 Application → Curr.output → Output curr.

Description

Shows the value currently calculated for the current output

User interface

3.59 to 23 mA

Terminal current
Navigation

 Application → Curr.output → Terminal curr.

Description

Shows the current value of the current output which is currently measured

User interface

0 to 30 mA

4 mA trim value

Navigation

Application → Curr.output → 4 mA trim value

Description

Enter the trim value for the 4 mA current output.
Note:
Simulation must be active.

User entry

3 to 5 mA

20 mA trim value

Navigation

Application → Curr.output → 20 mA trim value

Description

Enter the trim value for the 20 mA current output.
Note:
Simulation must be active.

User entry

Endress+Hauser

18 to 22 mA

149

Description of device parameters

3.4.5
Navigation

Cerabar PMP71B HART

"HART output" submenu
Application → HART output"Configuration" submenu
Navigation

Application → HART output → Configuration

HART addressNavigation

 Application → HART output → Configuration → HART address

Description

Define the HART address of the device.

User entry

0 to 63

Additional information• The measured value can only be transmitted via the current value if the address is set to"0". The current is fixed at 4.0 mA for all other addresses (Multidrop mode).
• Only addresses in the range 0 to 15 are permitted for a system according to HART 5.0.
• All addresses in the range 0 to 63 are permitted for a system with HART 6.0 and higher.

HART short tag

Navigation

 Application → HART output → Configuration → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Max. 8 characters: A to Z, 0 to 9 and certain special characters (e.g. punctuation marks, @,%).

Device tagNavigation

 Application → HART output → Configuration → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)150

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

No. of preambles



Navigation

 Application → HART output → Configuration → No. of preambles

Description

Defines the number of preambles in the HART telegram

User entry

5 to 20

Loop current modeNavigation

 Application → HART output → Configuration → Loop curr mode

Description

If Loop current mode is disabled, Multi-drop communication mode is activated. Multi-drop
is a HART digital communication mode where multiple devices may share the same pair ofwires for power and communications.
In this mode the output current is fixed.

Selection• Disable• Enable"HART output" submenu
Navigation

Application → HART output → HART output

Assign PV

Navigation

 Application → HART output → HART output → Assign PV

Description

Use this function to select a measured variable (HART device variable) for the primary
dynamic variable (PV).

Selection• Pressure• Scaled variable

Primary variable (PV)Navigation

Application → HART output → HART output → Primary var (PV)

Description

Shows the first HART value (PV).

Additional information

Endress+Hauser

151

Description of device parameters

Cerabar PMP71B HART

Assign SVNavigation

 Application → HART output → HART output → Assign SV

Description

Use this function to select a measured variable (HART device variable) for the secondary
dynamic variable (SV).

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Terminal current *• Terminal voltage *• Median of pressure signal *• Noise of pressure signal *• Signal noise detected *• Percent of range• Loop current• Not used

Additional information

Selection• Sensor pressure option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.
• Terminal current option
The terminal current is the read-back current on terminal block.
• Signal noise detected option
0 % - Signal noise is within the permissible range.
100 % - Signal noise is outside of the permissible range.
• Loop current option
The loop current is the output current set by the applied pressure.

Secondary variable (SV)Navigation

Application → HART output → HART output → Second.var(SV)Description

Shows the second HART value (SV).

Assign TV

Navigation

 Application → HART output → HART output → Assign TV

Description

Use this function to select a measured variable (HART device variable) for the tertiary(third) dynamic variable (TV).

*152

Visibility depends on order options or device settings

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Terminal current *• Terminal voltage *• Median of pressure signal *• Noise of pressure signal *• Signal noise detected *• Percent of range• Loop current• Not used

Additional information

Selection• Sensor pressure option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.
• Terminal current option
The terminal current is the read-back current on terminal block.
• Signal noise detected option
0 % - Signal noise is within the permissible range.
100 % - Signal noise is outside of the permissible range.
• Loop current option
The loop current is the output current set by the applied pressure.

Tertiary variable (TV)
Navigation

Application → HART output → HART output → Tertiary var(TV)Description

Shows the third HART value (TV).

Assign QVNavigation

 Application → HART output → HART output → Assign QV

Description

Use this function to select a measured variable (HART device variable) for the quaternary(fourth) dynamic variable (QV).

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Terminal current *• Terminal voltage *• Median of pressure signal *• Noise of pressure signal *• Signal noise detected **Visibility depends on order options or device settings

Endress+Hauser

153

Description of device parameters

Cerabar PMP71B HART

• Percent of range• Loop current• Not used
Additional information

Selection• Sensor pressure option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.
• Terminal current option
The terminal current is the read-back current on terminal block.
• Signal noise detected option
0 % - Signal noise is within the permissible range.
100 % - Signal noise is outside of the permissible range.
• Loop current option
The loop current is the output current set by the applied pressure.

Quaternary variable (QV)
Navigation

Application → HART output → HART output → Quaterna.var(QV)

Description

Shows the fourth value (QV).

"Burst configuration 1" submenu
Navigation

Application → HART output → Burst config. 1

Burst mode 1Navigation

Application → HART output → Burst config. 1 → Burst mode 1

Description

Use this function to select whether to activate the HART burst mode for burst message X.

Selection• Off
The measuring device transmits data only when requested by the HART master.
• On
The measuring device transmits data regularly without being requested.

Additional information

Selection• Off
The measuring device transmits data only when requested by the HART master.
• On
The measuring device transmits data regularly without being requested.

154

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Burst command 1

Navigation

Application → HART output → Burst config. 1 → Burst command 1

Description

Use this function to select the HART command that is sent to the HART master.

Selection• Primary variable (PV)• Loop Current and Percent of Range• Dynamic Variables• Device variables with status• Device variables• Additional device status

Additional information

Selection• Command 1
Read out the primary variable.
• Command 2
Read out the current and the main measured value as a percentage.
• Command 3
Read out the dynamic HART variables and the current.
• Command 9
Read out the dynamic HART variables including the related status.
• Command 33
Read out the dynamic HART variables including the related unit.
• Command 48
Read out the complete device diagnostics.
"Command 33" option
The HART device variables are defined via Command 107.
Commands• Information about the defined details of the command: HART specifications• The measured variables (HART device variables) are assigned to the dynamic
variables in the Output submenu.

Burst variable 0NavigationDescription

For HART command 9 and 33: select the HART device variable or the process variable.

Selection• Pressure• Scaled variable• Sensor temperature• Sensor pressure• Electronics temperature• Measured current *• Terminal voltage 1 *• Median of pressure signal *• Noise of pressure signal **Application → HART output → Burst config. 1 → Burst variable 0

Visibility depends on order options or device settings

Endress+Hauser

155

Description of device parameters

Cerabar PMP71B HART

• Signal noise detected *• Percent of range• Measured current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used
Additional information

Selection
If a burst message is not configured, the Not used option is set.

Burst variable 1Navigation

Application → HART output → Burst config. 1 → Burst variable 1

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection

Please refer to the Burst variable 0 parameter (→  155).

Burst variable 2

Navigation

Application → HART output → Burst config. 1 → Burst variable 2

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection

Please refer to the Burst variable 0 parameter (→  155).

Burst variable 3

Navigation

Application → HART output → Burst config. 1 → Burst variable 3

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection

Please refer to the Burst variable 0 parameter (→  155).

Burst variable 4NavigationDescription

For HART command 9: select the HART device variable or the process variable.

*

156

Application → HART output → Burst config. 1 → Burst variable 4

Visibility depends on order options or device settings

Endress+Hauser

Cerabar PMP71B HART

Selection

Description of device parameters

Please refer to the Burst variable 0 parameter (→  155).

Burst variable 5Navigation

Application → HART output → Burst config. 1 → Burst variable 5

Description

For HART command 9: select the HART device variable or the process variable.

Selection

Please refer to the Burst variable 0 parameter (→  155).

Burst variable 6

Navigation

Application → HART output → Burst config. 1 → Burst variable 6

Description

For HART command 9: select the HART device variable or the process variable.

Selection

Please refer to the Burst variable 0 parameter (→  155).

Burst variable 7

Navigation

Application → HART output → Burst config. 1 → Burst variable 7

Description

For HART command 9: select the HART device variable or the process variable.

Selection

Please refer to the Burst variable 0 parameter (→  155).

Burst trigger modeNavigationDescription

Use this function to select the event that triggers burst message X.

Selection• Continuous• Window *• Rising *• Falling *• On change*

Application → HART output → Burst config. 1 → Trigger mode

Visibility depends on order options or device settings

Endress+Hauser

157

Description of device parameters

Additional information

Cerabar PMP71B HART

Selection• Continuous
The message is sent continuously, at least at intervals corresponding to the time frame
specified in the Burst min period parameter (→  158).
• Window
The message is sent if the specified measured value has changed by the value in the
Burst trigger level parameter (→  158).
• Rising
The message is sent if the specified measured value exceeds the value in the Burst
trigger level parameter (→  158).
• Falling
The message is sent if the specified measured value drops below the value in the Burst
trigger level parameter (→  158).
• On change
The message is sent if a measured value changes in the burst message.

Burst trigger level

Navigation

Application → HART output → Burst config. 1 → Trigger level

Description

Use this function to enter the burst trigger value.

User entry

Signed floating-point number

Additional information

Description
Together with the option selected in the Burst trigger mode parameter (→  157) the
burst trigger value determines the time of burst message X.

Min. update period

Navigation

Application → HART output → Burst config. 1 → Min. upd. per.

Description

Use this function to enter the minimum time span between two burst commands of burst
message X.

User entry

Positive integer

Max. update period

NavigationDescription

Use this function to enter the maximum time span between two burst commands of burst
message X.

User entry

Positive integer

158

Application → HART output → Burst config. 1 → Max. upd. per.

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters"Information" submenu
Navigation

Application → HART output → Information

Device ID
Navigation

 Application → HART output → Information → Device ID

Description

Use this function to view the device ID for identifying the measuring device in a HART
network.

User interface

6-digit hexadecimal number

Additional information

Description
In addition to the device type and manufacturer ID, the device ID is part of the unique
ID. Each HART device is uniquely identified by the unique device ID.

Device type
Navigation

 Application → HART output → Information → Device type

Description

Displays the device type with which the measuring device is registered with the HART
Communication Foundation.

User interface

2-digit hexadecimal number

Factory setting

0x54

Factory setting

0x5A

Additional information

Description
The device type is specified by the manufacturer. It is needed to assign the appropriate
device description file (DD) to the device.

Device revision
Navigation

 Application → HART output → Information → Device revision

Description

Displays the device revision with which the device is registered with the HART
Communication Foundation.

User interface

2-digit hexadecimal number

Endress+Hauser

159

Description of device parameters

Additional information

Cerabar PMP71B HART

Description
The device revision is needed to assign the appropriate device description file (DD) tothe device.

HART short tag

Navigation

 Application → HART output → Information → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Max. 8 characters: A to Z, 0 to 9 and certain special characters (e.g. punctuation marks, @,%).

HART revision
Navigation

 Application → HART output → Information → HART revision

Description

Shows the HART revision of the device.

HART descriptor

Navigation

 Application → HART output → Information → HART descriptor

Description

Description for the measuring point.

User entry

Character string comprising numbers, letters and special characters (16)

HART message

Navigation

 Application → HART output → Information → HART message

Description

A HART message which is sent via the HART protocol when requested by the master.

User entry

Character string comprising numbers, letters and special characters (32)160

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

HART date code

Navigation

 Application → HART output → Information → HART date code

Description

Date of the last configuration change

User entry

Character string comprising numbers, letters and special characters (10)Additional information

Date format: YYYY-MM-DD
Make sure you adhere to this format when entering the date. Otherwise errors may
occur in individual HART commands.

Endress+Hauser

161

Description of device parameters

Cerabar PMP71B HART

3.5

"System" menu

Navigation

3.5.1



System"Device management" submenu

Navigation



System → Device manag.

Device tagNavigation

 System → Device manag. → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)

Locking status
Navigation

 System → Device manag. → Locking status

Description

Displays the active write protection.

User interface• Hardware locked• Safety locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.
Selection
Function scope of the "Locking status" parameter

162

Options

Description

None

The access status displayed in the Access status display parameter applies. Only
appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the main electronics module.
This prevents write access to the parameters (e.g. via the local display or operatingtool).

Temporarily locked

Write access to the parameters is temporarily locked due to device-internal
processing (e.g. data upload/download, reset). Once the internal processing has
been completed, the parameters can be changed once again.

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Configuration counter
Navigation

 System → Device manag. → Config. counter

Description

Displays the counter for changes to the device parameters.
Additional information:- If the value for a static parameter is changed when optimizing or configuring theparameter, the counter is incremented by 1. This is to enable tracking different parameterversions.
- When multiple parameters are changed simultaneously, e.g. when loading parameters
into the device from an external source such as FieldCare, the counter may display a higher
value. The counter cannot be reset, nor is it reset to a default value on performing a device
reset.
- Once the counter has reached the value 65535, it restarts at 0.

User interface

0 to 65 535

Reset device

Navigation

 System → Device manag. → Reset device

Description

Use this function to choose whether to reset the device configuration - either entirely or inpart - to a defined state.

Selection• Cancel• To factory defaults *• To delivery settings *• Restart device

Additional information

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

To factory defaults

Every parameter is reset to its factory setting.

To delivery settings

Every parameter for which a customer-specific default setting was ordered is resetto this customer-specific value. All other parameters are reset to the factory
setting.

 This option is not visible if no customer-specific settings have been ordered.
Restart device*

The restart resets every parameter whose data are in the volatile memory (RAM)to the factory setting (e.g. measured value data). The device configuration remains
unchanged.

Visibility depends on order options or device settings

Endress+Hauser

163

Description of device parameters

Cerabar PMP71B HART

3.5.2

"User management" submenu

Navigation



System → User manag.

User role
Navigation

 System → User manag. → User role

Description

Displays the access authorization to the parameters via the operating tool.

User interface• Operator• Maintenance• Expert

Additional information

Description
Access authorization can be modified via the Enter access code parameter.
If additional write protection is active, this restricts the current access authorizationeven further.
User interface
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.

Password
Navigation



System → User manag. → Password

Description

Enter the password for the "Maintenance" user role to get access to the functionality of thisrole.

User entry

Character string comprising numbers, letters and special characters (16)Enter access code

NavigationDescription

Use this function to enter the user-specific release code to remove parameter write
protection in the operating tool.

User entry

0 to 9 999

164

System → User manag. → Ent. access code

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Status password entry
Navigation

 System → User manag. → Status pw entry

Description

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

New passwordNavigation

 System → User manag. → New password

Description

Define the new "Maintenance" password.
A new password is valid after it has been confirmed within the "Confirm new password"
parameter.
Any valid password consists of 4 to 16 characters and can contain letters and numbers.

User entry

Character string comprising numbers, letters and special characters (16)

Confirm new passwordNavigation

 System → User manag. → Confirm password

Description

Enter the new password again to confirm.

User entry

Character string comprising numbers, letters and special characters (16)

Old passwordNavigation

 System → User manag. → Old password

Description

Enter the current password, to subsequently change the existing password.

User entry

Character string comprising numbers, letters and special characters (16)Endress+Hauser

165

Description of device parameters

Cerabar PMP71B HART

Reset password
Navigation



System → User manag. → Reset password

Description

Enter a code to reset the current "Maintenance" password.
The code is delivered by your local support.

User entry

Character string comprising numbers, letters and special characters (16)3.5.3"Bluetooth configuration" submenu

Navigation



System → Bluetooth conf.

Bluetooth activation
Navigation

 System → Bluetooth conf. → Bluetooth active

Description

If Bluetooth is deactivated, it can only be reactivated via the display or the operating tool.
Reactivating via the SmartBlue app is not possible.

Selection• Disable• Enable

3.5.4"Display" submenu

Navigation



System → Display

Language
Navigation

 System → Display → Language

Prerequisite

A local display is provided.

Description

Use this function to select the configured language on the local display.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands

166

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• Bahasa Indonesia• tiếng Việt (Vietnamese)• čeština (Czech)
Factory setting

English (alternatively, the ordered language is preset in the device)

Format display
Navigation

 System → Display → Format display

Prerequisite

A local display is provided.

Description

Use this function to select how the measured value is shown on the local display.

Selection• 1 value, max. size
• 1 bargraph + 1 value• 2 values

Additional information

Description
The display format (size, bar graph etc.) and number of measured values displayed
simultaneously (1 to 4) can be configured. This setting only applies to normal operation.
• The Value 1 display parameter (→  167)...Value 8 display parameter Value 4
display parameter (→  169) are used to specify which measured values areshown on the local display and in what order.
• If more measured values are specified than the display mode selected permits, thenthe values alternate on the device display. The display time until the next change isconfigured using the Display interval parameter.

Value 1 display

Navigation

 System → Display → Value 1 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• Pressure• Scaled variable• Current output• Sensor temperature• Percent of range

Endress+Hauser

167

Description of device parameters

Additional information

Cerabar PMP71B HART

Description
If several measured values are displayed at once, the measured value selected here will be
the first value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  167) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu.

Value 2 display

Navigation

 System → Display → Value 2 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• None• Pressure• Scaled variable• Current output• Sensor temperature• Percent of range

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the second value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  167) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu.

Value 3 display

Navigation

 System → Display → Value 3 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• None• Pressure• Scaled variable• Current output• Sensor temperature• Percent of range

168

Endress+Hauser

Cerabar PMP71B HART

Additional information

Description of device parameters

Description
If several measured values are displayed at once, the measured value selected here will be
the third value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  167) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu.

Value 4 display

Navigation

 System → Display → Value 4 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• None• Pressure• Scaled variable• Current output• Sensor temperature• Percent of range

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the fourth value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  167) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu.

Contrast display
Navigation

 System → Display → Contrast display

Description

Adjust local display contrast setting to ambient conditions (e.g. lighting or reading angle)

User entry

20 to 80 %Factory setting

Depends on the display

Endress+Hauser

169

Description of device parameters

Additional information

Cerabar PMP71B HART

Set the contrast via the push-buttons:• Weaker: Press the and buttons simultaneously• Stronger: Press the and buttons simultaneously
E

E

3.5.5

"Geolocation" submenu

Navigation



System → Geolocation

Process Unit Tag

Navigation



System → Geolocation → Process Unit Tag

Description

Enter the process unit in which the device is installed.

User entry

Character string comprising numbers, letters and special characters (32)

Location Description



Navigation



System → Geolocation → Location Descr.

Description

Use this function to enter a description of the location so that the device can be located in
the plant.

User entry

Character string comprising numbers, letters and special characters (32)Longitude

Navigation



System → Geolocation → Longitude

Description

Use this function to enter the longitude coordinates that describe the device location.

User entry–180 to 180 °LatitudeNavigationDescription

Use this function to enter the latitude coordinates that describe the device location.

170

System → Geolocation → Latitude

Endress+Hauser

Cerabar PMP71B HART

User entry

Description of device parameters–90 to 90 °AltitudeNavigation



System → Geolocation → Altitude

Description

Use this function to enter the altitude data that describe the device location.

User entry

Signed floating-point number

Location methodNavigation



System → Geolocation → Location method

Description

Use this function to select the data format for specifying the geographic location. The
codes for specifying the location are based on the US National Marine Electronics
Association (NMEA) Standard NMEA 0183.

Selection• No fix• GPS or Standard Positioning Service fix• Differential GPS fix• Precise positioning service (PPS) fix• Real Time Kinetic (RTK) fixed solution• Real Time Kinetic (RTK) float solution• Estimated dead reckoning• Manual input mode
• Simulation Mode

3.5.6
Navigation"Information" submenu


System → Information

Device name
Navigation

 System → Information → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

Endress+Hauser

171

Description of device parameters

Cerabar PMP71B HART

Manufacturer
Navigation

 System → Information → Manufacturer

User interface

Character string comprising numbers, letters and special characters

Serial number



Navigation

 System → Information → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

Order code

Navigation

 System → Information → Order code

Description

Shows the device order code.

User interface

Character string composed of letters, numbers and certain punctuation marks (e.g. /).

Factory setting–

Additional information

Description
The order code is generated from the extended order code through a process of reversible
transformation. The extended order code indicates the attributes for all the device featuresin the product structure. The device features are not directly readable from the order code.
Uses of the order code
• To order an identical spare device.
• To identify the device quickly and easily, e.g. when contacting Endress+Hauser.

172

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

Firmware version
Navigation

 System → Information → Firmware version

Description

Displays the device firmware version that is installed.

User interface

Character string in the format xx.yy.zz

Additional information

User interface
The Firmware version is also located:• On the title page of the Operating instructions• On the transmitter nameplate

Hardware version
Navigation

 System → Information → Hardware version

User interface

Character string comprising numbers, letters and special characters

Extended order code 1Navigation



System → Information → Ext. order cd. 1

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string

Factory setting–

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.

Extended order code 2Navigation



System → Information → Ext. order cd. 2

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface
Endress+Hauser

Character string173

Description of device parameters

Factory setting

Cerabar PMP71B HART–

Extended order code 3

Navigation



System → Information → Ext. order cd. 3

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface

Character string

Factory setting–

XML build number
Navigation

 System → Information → XML build no.

User interface

Positive integer

Checksum
Navigation

 System → Information → Checksum

Description

Checksum for Firmware version.

User interface

Positive integer

174

Endress+Hauser

Cerabar PMP71B HART

Description of device parameters

3.5.7"Software configuration" submenu

Navigation



System → Softw. config.

CRC device configuration
Navigation

 System → Softw. config. → CRC device conf.

Description

CRC device configuration based on current settings of safety relevant parameters.
The CRC device configuration is unique and can be used to detect changes in safety
relevant parameter settings.

User interface

0 to 65 535

Stored CRC device configuration
Navigation

 System → Softw. config. → Stored CRC conf.

Description

Stored CRC after the last safety lock. Factory delivery is 65535 means that the device has
not yet been safety locked.

User interface

0 to 65 535

Timestamp stored CRC device config.
Navigation

 System → Softw. config. → Time stored CRC

Description

Gives the time stamp when the CRC was last stored following completion of the safety lockwizard.

User interface

Character string comprising numbers, letters and special characters

Activate SW option



Navigation

 System → Softw. config. → Activate SW opt.

Description

Use this function to enter an activation code to enable an additional, ordered software
option.

User entry

Max. 10-digit string of numbers.

Factory setting

Depends on the software option ordered

Endress+Hauser

175

Description of device parameters

Additional information

Cerabar PMP71B HART

Description
If a measuring device was ordered with an additional software option, the activation codeis programmed in the device at the factory.
User entry
To activate a software option subsequently, please contact your Endress+Hauser salesorganization.
NOTE!
The activation code is linked to the serial number of the measuring device and varies
according to the device and software option.
If an incorrect or invalid code is entered, this results in the loss of software options thathave already been activated.
▸ Before you enter a new activation code, make a note of the current activation code fromthe parameter protocol.
▸ Enter the new activation code provided by Endress+Hauser when the new software
option was ordered.
▸ If the code entered is incorrect or invalid, enter the old activation code from theparameter protocol.
▸ Have the Endress+Hauser sales organization check the new activation code
remembering to specify the serial number or ask for the code again.
Example for a software option
Order code for "Application package", option EA "Extended HistoROM"

Software option overview
Navigation

 System → Softw. config. → SW option overv.

Description

Shows all enabled software options

User interface

• SIL
• WHG
• Heartbeat Verification• Heartbeat Monitoring

176

Endress+Hauser

Cerabar PMP71B HART

Index

Index
Symbols(High) Is safety function triggered? (Parameter) . . . . 100(Low) Is safety function triggered? (Parameter) . . . . 101

0…94 mA trim value (Parameter) . . . . . . . . . . . . . . . . . . 149
20 mA trim value (Parameter) . . . . . . . . . . . . . . . . . 149
500 Diagnostic behavior (Parameter) . . . . . . . . . 77, 122
500 Event category (Parameter) . . . . . . . . . . . . 77, 122
500 Process alert pressure (Parameter) . . . . . . . 75, 119
501 Diagnostic behavior (Parameter) . . . . . . . . . 78, 122
501 Event category (Parameter) . . . . . . . . . . . . 79, 123
501 Process alert scaled variable (Parameter) . . . 77, 119
502 Diagnostic behavior (Parameter) . . . . . . . . . 80, 123
502 Event category (Parameter) . . . . . . . . . . . . 80, 123
806 Diagnostic behavior (Parameter) 74, 121, 124, 127
806 Event category (Parameter) . . . . . 74, 121, 124, 127
806 Event delay (Parameter) . . . . . . . . . . . 74, 117, 121
822 Diagnostic behavior (Parameter) . . . . . . . . 124, 127
822 Event category (Parameter) . . . . . . . . . . . 125, 128
841 Event category (Parameter) . . . . . . . . . . . 125, 128
900 Diagnostic behavior (Parameter) . . . . . 70, 126, 129
900 Event category (Parameter) . . . . . . . . . 69, 126, 129
906 Diagnostic behavior (Parameter) . . . . . . . . 126, 129
906 Event category (Parameter) . . . . . . . . . 70, 126, 129A

Activate SW option (Parameter) . . . . . . . . . . . . . . . 175
Activate table (Parameter) . . . . . . . . . . . . . . . . . . . . 143
Active diagnostics (Parameter) . . . . . . . . . . . . 103, 106
Active diagnostics (Submenu) . . . . . . . . . . . . . . . . . 106
Altitude (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 171
Analog path integrity (Parameter) . . . . . . . . . . . . . . . 58
Application (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . 131
Assign HART variables? (Parameter) . . . . . . . . . . 50, 52
Assign PV (Parameter) . . . . . . 38, 52, 90, 140, 147, 151
Assign QV (Parameter) . . . . . . . . . . . . . . . . . . . 54, 153
Assign SV (Parameter) . . . . . . . . . . . . . . . . . . . 53, 152
Assign TV (Parameter) . . . . . . . . . . . . . . . . . . . 53, 152

B

Baseline build process (Parameter) . . . . . . . . . . . . 64, 72
Baseline is available (Parameter) . . . . . . . . . . . . . . . . 65
Baseline Signal Control Line (Parameter) . . . . . . . . . . 66
Baseline Signal Lower Control Line (Parameter) . . . . . 67
Baseline Signal Noise Control Line (Parameter) . . . . . 68
Baseline Signal Noise Lower Control Line (Parameter) 68
Baseline Signal Noise Minimum (Parameter) . . . . . . . 69
Baseline Signal Noise Upper Control Line (Parameter) 68
Baseline Signal Upper Control Line (Parameter) . . . . . 66
Baseline status (Parameter) . . . . . . . . . . . . . . . . 71, 116
Basic settings (Submenu) . . . . . . . . . . . . . . . . . . . . . 135
Bluetooth activation (Parameter) . . . . . . . . . . . . . . . 166
Bluetooth configuration (Submenu) . . . . . . . . . . . . . 166
Burst command 1 (Parameter) . . . . . . . . . . . . . . . . . 155
Burst configuration 1 (Submenu) . . . . . . . . . . . . . . . 154
Endress+Hauser

Burst mode 1 (Parameter) . . . . . . . . . . . . . . . . . . . . 154
Burst trigger level (Parameter) . . . . . . . . . . . . . . . . . 158
Burst trigger mode (Parameter) . . . . . . . . . . . . . . . . 157
Burst variable 0 (Parameter) . . . . . . . . . . . . . . . . . . 155
Burst variable 1 (Parameter) . . . . . . . . . . . . . . . . . . 156
Burst variable 2 (Parameter) . . . . . . . . . . . . . . . . . . 156
Burst variable 3 (Parameter) . . . . . . . . . . . . . . . . . . 156
Burst variable 4 (Parameter) . . . . . . . . . . . . . . . . . . 156
Burst variable 5 (Parameter) . . . . . . . . . . . . . . . . . . 157
Burst variable 6 (Parameter) . . . . . . . . . . . . . . . . . . 157
Burst variable 7 (Parameter) . . . . . . . . . . . . . . . . . . 157

C

Calibration offset (Parameter) . . . . . . . . . . . . . . . . . 136
Character test string (Parameter) . . . . . . . . . . . . . . . . 83
Checksum (Parameter) . . . . . . . . . . . . . . . . . . . . . . 174
Clamping voltage lower threshold (Parameter) . . . . . 116
Clamping voltage upper threshold (Parameter) . . . . . 116
Clear event list (Parameter) . . . . . . . . . . . . . . . . . . . 108
Commissioning (Wizard) . . . . . . . . . . . . . . . . . . . . . . 34
Configuration (Submenu) . . . . . . . . . . . . . . . . 122, 150
Configuration counter (Parameter) . . . . . . . . 87, 98, 163
Confirm (Parameter) . . . . . . . . . 83, 85, 88, 89, 91, 100
Confirm new password (Parameter) . . . . . . . . . . . . . 165
Confirmation (Wizard) . . . . . . . . . . . . . . . 87, 88, 89, 91
Contrast display (Parameter) . . . . . . . . . . . . . . . . . . 169
Counter Baseline creation SSD (Parameter) . . . . 65, 118
Counter limit overruns sensor Pmax (Parameter) . . . 111
Counter limit overruns sensor Tmax (Parameter) . . . 112
Counter limit underruns sensor Pmin (Parameter) . . 109
Counter limit underruns sensor Tmin (Parameter) . . 110
Counter overruns of user limit Pmax (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76, 111
Counter overruns of user limit Tmax (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80, 112
Counter underruns of user limit Pmin (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76, 109
Counter underruns of user limit Tmin (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80, 110
CRC device configuration (Parameter) . . . . . . 86, 97, 175
Current Baseline noise (Parameter) . . . . . . . . . . . . . . 68
Current Baseline signal (Parameter) . . . . . . . . . . . . . . 66
Current deviation 1 (Parameter) . . . . . . . . . . . . . . . 102
Current deviation 2 (Parameter) . . . . . . . . . . . . . . . 102
Current output (Submenu) . . . . . . . . . . . . . . . . . . . . 147
Current range output (Parameter) . . . . . 49, 51, 90, 147

D

Damping (Parameter) . . . . . . . . . . . . . . . . . 38, 88, 136
Date/time (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 93
Date/time Heartbeat Verification (Parameter) . . 60, 114
Date/time proof test (Parameter) . . . . . . . . . . . . . . . 104
Decimal places pressure (Parameter) . . . . . . . . . . . . 131
Decimal places scaled variable (Parameter) . . . . . . . 133
Decision of inspector (Parameter) . . . . . . . . . . . . . . 103
Description of device parameters . . . . . . . . . . . . . . . . 31
177

Index

Device ID (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 159
Device identification (Wizard) . . . . . . . . . . . . 34, 36, 37
Device management (Submenu) . . . . . . . . . . . . . . . . 162
Device name (Parameter) . . . . . . . . . . . . 34, 84, 96, 171
Device revision (Parameter) . . . . . . . . . . . . . . . . . . . 159
Device tag (Parameter) . . . . . . . . . . 34, 84, 95, 150, 162
Device type (Parameter) . . . . . . . . . . . . . . . . . . . . . 159
Diagnostic event simulation (Parameter) . . . . . . . . . 113
Diagnostic settings (Submenu) . . . . . . . . . . . . . . . . . 118
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . 106
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 166
Document
Explanation of the structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using the document . . . . . . . . . . . . . . . . . . . . . . . . 4
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

EElectronics temperature (Parameter) . . . . . . . . . . . . 135
Enter access code (Parameter) . . . . . . . . . . . . . . . . . 164
Enter safety locking code (Parameter) . . . . . . . . . 83, 91
Enter safety unlocking code (Parameter) . . . . . . . . . . 81
Event logbook (Submenu) . . . . . . . . . . . . . . . . . . . . 108
Extended order code 1 (Parameter) . . . . . . . . . . 35, 173
Extended order code 2 (Parameter) . . . . . . . . . . 35, 173
Extended order code 3 (Parameter) . . . . . . . . . . 35, 174

F

Failure behavior current output (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49, 51, 89, 148
Failure current (high alarm) (Parameter) . . . . . . . . . . 99
Failure current (Parameter) . . . . . . . . . . . . . 49, 52, 148
Filter options (Parameter) . . . . . . . . . . . . . . . . . . . . 108
Finish (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . 93, 103
Firmware version (Parameter) . . . . . . . . . . . 85, 96, 173
Format display (Parameter) . . . . . . . . . . . . . . . . . . . 167
Free text (Parameter) . . . . . . . . . . . . . . . . 41, 133, 141
Function
see Parameter

G

Geolocation (Submenu) . . . . . . . . . . . . . . . . . . . . . . 170
Guidance (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . 31, 34

H

Hardware version (Parameter) . . . . . . . . . . . 85, 97, 173
HART address (Parameter) . . . . . . . . . . . . . . . . 38, 150
HART date code (Parameter) . . . . . . . . . . . . . . . 37, 161
HART descriptor (Parameter) . . . . . . . . . . . . . . . 37, 160
HART message (Parameter) . . . . . . . . . . . . . . . . 37, 160
HART output (Submenu) . . . . . . . . . . . . . . . . . 150, 151
HART revision (Parameter) . . . . . . . . . . . . . . . . . . . 160
HART short tag (Parameter) . . . . . . . . . . . 37, 150, 160
Heartbeat Technology (Submenu) . . . . . . . . . . . . . . 114
Heartbeat Verification (Parameter) . . . . . . . . . . . . . . 55
Heartbeat Verification (Submenu) . . . . . . . . . . . . . . 114
178

Cerabar PMP71B HART

Heartbeat Verification (Wizard) . . . . . . . . . . . . . . . . . 55
High alert value (Parameter) . 76, 78, 79, 119, 120, 121
HP/LP swap (Parameter) . . . . . . . . . . . . . . . . . . . . . . 87

I

I have read the warning texts. (Parameter) . . . 71, 81, 94
Information (Submenu) . . . . . . . . . . . . . . . . . . 159, 171
Inspector (Parameter) . . . . . . . . . . . . . . . . . 59, 93, 104

L

Language (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 166
Latitude (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 170
Location (Parameter) . . . . . . . . . . . . . . . . . . 59, 93, 104
Location Description (Parameter) . . . . . . . . . . . . . . . 170
Location method (Parameter) . . . . . . . . . . . . . . . . . 171
Locking (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . . . 91
Locking status (Parameter) . . . . . . . . . . 36, 81, 92, 162
Longitude (Parameter) . . . . . . . . . . . . . . . . . . . . . . 170
Loop current mode (Parameter) . . . . . . . . . . 50, 52, 151
Loop diagnostics (Parameter) . . . . . . . . . . . . 57, 75, 116
Loop diagnostics (Submenu) . . . . . . . . . . . . . . . . . . 115
Low alert value (Parameter) . . . . . . 75, 78, 79, 119, 120
Low cutoff (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 89
Lower Range Limit (Parameter) . . . . . . . 44, 46, 48, 139
Lower range value output (Parameter). . . . . . . . . . . . . . . . 45, 48, 51, 90, 142, 145, 146, 148
Lower sensor trim (Parameter) . . . . . . . . . . . . . . . . 138
Lower sensor trim measured value (Parameter) . . . . 137

M

Mainboard module (Wizard) . . . . . . . . . . . . . . . . . . . 55
Manufacturer (Parameter) . . . . . . . . . . . . . . . . . . . . 172
Max. update period (Parameter) . . . . . . . . . . . . . . . 158
Maximum electronics temperature (Parameter) . . . . 112
Maximum sensor temperature (Parameter) . . . . . . . 111
Maximum terminal voltage (Parameter) . . . . . . . . . 112
Measured values (Submenu) . . . . . . . . . . . . . . . . . . 134
Measurement adjustments (Wizard) . . . . . 38, 39, 40, 42
Measuring mode current output (Parameter) . . . 89, 147
Measuring units (Submenu) . . . . . . . . . . . . . . . . . . . 131
Membrane integrity (Parameter) . . . . . . . . . . . . . . . . 57
Menu
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . 131
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Guidance . . . . . . . . . . . . . . . . . . . . . . . . . . . . 31, 34
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 162
Min. update period (Parameter) . . . . . . . . . . . . . . . . 158
Minimum electronics temperature (Parameter) . . . . 110
Minimum sensor temperature (Parameter) . . . . . . . 110
Minimum span (Parameter) . . . . . . . . . . 44, 46, 48, 140
Minimum terminal voltage (Parameter) . . . . . . . . . . 110
Minimum/maximum values (Submenu) . . . . . . . . . . 109

N

New password (Parameter) . . . . . . . . . . . . . . . . . . . 165
No. of preambles (Parameter) . . . . . . . . . . . . . . . . . 151
Notes (Parameter) . . . . . . . . . . . . . . . . . 60, 93, 95, 104

O

Old password (Parameter) . . . . . . . . . . . . . . . . . . . . 165
Endress+Hauser

Cerabar PMP71B HART

Operating time (Parameter) . . . . . . . . . . . . . 86, 98, 108
Operating time (Verification) (Parameter) . . . . . 60, 114
Operating time from restart (Parameter) . . . . . . . . . 107
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . 172
Output current (Parameter) . . . . . . . . . . . . . . . . 55, 149
Output current transfer function (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43, 88, 135
Output settings (Wizard) . . . . 43, 44, 45, 46, 48, 50, 52

PParameter
Structure of a parameter description . . . . . . . . . . . . 4
Password (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 164
Plant operator (Parameter) . . . . . . . . . . . . . 60, 94, 105
Preparation (Wizard) . . . . . . . . . . . . . 81, 82, 83, 84, 86
Preparation Proof test (Wizard) . . . . . . . . 94, 95, 97, 99
Pressure (Parameter) . . . . . . . . . . 43, 45, 134, 142, 144
Pressure 1 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 101
Pressure 2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 102
Pressure max (Parameter) . . . . . . . . . . . . . . . . . . . . 111
Pressure min (Parameter) . . . . . . . . . . . . . . . . . . . . 109
Pressure unit (Parameter) . . . . . . . . . . . . . . 39, 40, 131
Pressure value 1 (Parameter) . . . . . . . 47, 143, 144, 146
Pressure value 2 (Parameter) . . . . . . . 47, 143, 145, 146
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . 107
Primary variable (PV) (Parameter) . . . . . . . . . . . . . . 151
Process (Submenu) . . . . . . . . . . . . . . . . . . . . . 124, 127
Process Unit Tag (Parameter) . . . . . . . . . . . . . . . . . 170
Process variable output current (Parameter) . . . . . . . . 50
Proof test (Wizard) . . . . . . . . . . . . . . . . . . . . . . 94, 101
Proof test via Bluetooth allowed? (Parameter) . . . . . . 82
Properties (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 118

Q

Quaternary variable (QV) (Parameter) . . . . . . . . . . . 154

R

RAM check (Parameter) . . . . . . . . . . . . . . . . . . . . . . 56
Rebuild baseline (Parameter) . . . . . . . . . . . . . . . . . . 115
Reset device (Parameter) . . . . . . . . . . . . . . . . . . . . . 163
Reset password (Parameter) . . . . . . . . . . . . . . . . . . 166
Reset user defined counters P and T (Parameter) . . . 111
Resistance Baseline (Parameter) . . . . . . . . . . . . . . . . 72
Resistance previous Baseline (Parameter) . . . . . . . . . . 73
ROM check (Parameter) . . . . . . . . . . . . . . . . . . . . . . 56

S

Safety function? (Parameter) . . . . . . . . . . . . . . . . . . . 99
Safety mode (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . 81
Sample rate (Parameter) . . . . . . . . . . . . . . . . . . . . . . 65
Save protocol? (Parameter) . . . . . . . . . . . . . . . . . 59, 92
Scaled variable (Parameter) . . . . . . . . . . . . 45, 134, 144
Scaled variable (Submenu) . . . . . . . . . . . . . . . . . . . . 140
Scaled variable transfer function (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43, 46, 142
Scaled variable unit (Parameter) . . . . . . . . . 40, 132, 141
Scaled variable value 1 (Parameter) . . . . . . . . . . 47, 143
Scaled variable value 2 (Parameter) . . . . . . . . . . 47, 143
Secondary variable (SV) (Parameter) . . . . . . . . . . . . 152
Select test procedure (Parameter) . . . . . . . . . . . . . . . 99
Endress+Hauser

Index

Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Sensor calibration (Submenu) . . . . . . . . . . . . . . . . . 136
Sensor integrity (Parameter) . . . . . . . . . . . . . . . . . . . 57
Sensor limits (Submenu) . . . . . . . . . . . . . . . . . . . . . 139
Sensor pressure (Parameter) . . . . . . . . . . . . . . . . . . 134
Sensor pressure range behavior (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 88, 125, 128
Sensor temperature (Parameter) . . . . . . . . . . . . . . . 134
Sensor temperature lower range limit (Parameter) . . 140
Sensor temperature upper range limit (Parameter) . . 140
Sensor Trim Reset (Parameter) . . . . . . . . . . . . . . . . 137
Sensor/membrane integrity (Parameter) . . . . . . . . . . 58
Serial number (Parameter) . . . . . . . . . . . 34, 85, 96, 172
Setpoint 1 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 101
Setpoint 2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 102
Signal maximum value (Parameter) . . . . . . . . . . . . . . 67
Signal minimum value (Parameter) . . . . . . . . . . . . . . 67
Signal noise maximum value (Parameter) . . . . . . . . . . 69
Signal noise minimum value (Parameter) . . . . . . . . . . 69
Signal noise status (Parameter) . . . . . . . . . . 64, 67, 118
Signal status (Parameter) . . . . . . . . . . . . . . . 64, 66, 117
Simulation (Parameter) . . . . . . . . . . . . . . . . . . . . . . 113
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 113
Simulation and check failure current (Wizard) . . . . . . 99
Software configuration (Submenu) . . . . . . . . . . . . . . 175
Software integrity (Parameter) . . . . . . . . . . . . . . . . . 56
Software option overview (Parameter) . . . . . . . . . . . 176
Span (Parameter) . . . . . . . . . . . . . . . . . . . . . . 144, 146
SSD Monitoring delay time (Parameter) . . . . . . . 69, 119
SSD Out of range delay time (Parameter) . . . . . . 70, 118
SSD: Statistical Sensor Diagnostics (Parameter) . . 71, 117
Statistical Sensor Diagnostics (Parameter) . . . . . . . . . 58
Statistical Sensor Diagnostics (Submenu) . . . . . . . . . 117
Status (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 115
Status password entry (Parameter) . . . . . . . . . . . . . 165
Status summary (Parameter) . . . . . . . . . . . . . 61, 62, 63
Stored CRC device configuration (Parameter) 86, 97, 175
Submenu
Active diagnostics . . . . . . . . . . . . . . . . . . . . . . . 106
Basic settings . . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Bluetooth configuration . . . . . . . . . . . . . . . . . . . 166
Burst configuration 1 . . . . . . . . . . . . . . . . . . . . . 154
Configuration . . . . . . . . . . . . . . . . . . . . . . . 122, 150
Current output . . . . . . . . . . . . . . . . . . . . . . . . . . 147
Device management . . . . . . . . . . . . . . . . . . . . . . 162
Diagnostic settings . . . . . . . . . . . . . . . . . . . . . . . 118
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . 108
Geolocation . . . . . . . . . . . . . . . . . . . . . . . . . . . . 170
HART output . . . . . . . . . . . . . . . . . . . . . . . 150, 151
Heartbeat Technology . . . . . . . . . . . . . . . . . . . . 114
Heartbeat Verification . . . . . . . . . . . . . . . . . . . . 114
Information . . . . . . . . . . . . . . . . . . . . . . . . 159, 171
Loop diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . 115
Measured values . . . . . . . . . . . . . . . . . . . . . . . . 134
Measuring units . . . . . . . . . . . . . . . . . . . . . . . . . 131
Minimum/maximum values . . . . . . . . . . . . . . . . 109
Process . . . . . . . . . . . . . . . . . . . . . . . . . . . 124, 127
Properties . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 118
179

Index

Scaled variable . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Sensor calibration . . . . . . . . . . . . . . . . . . . . . . . . 136
Sensor limits . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 113
Software configuration . . . . . . . . . . . . . . . . . . . . 175
Statistical Sensor Diagnostics . . . . . . . . . . . . . . . 117
User management . . . . . . . . . . . . . . . . . . . . . . . 164
Wet calibration . . . . . . . . . . . . . . . . . . . . . 144, 145
Summary (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 103
Supply voltage Baseline (Parameter) . . . . . . . . . . . . . 72
Supply voltage previous Baseline (Parameter) . . . . . . . 73
System (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 162
System status (Parameter) . . . . . . . . . . . . . . 55, 64, 117T

Cerabar PMP71B HART

Mainboard module . . . . . . . . . . . . . . . . . . . . . . . . 55
Measurement adjustments . . . . . . . . . 38, 39, 40, 42
Output settings . . . . . . . . . 43, 44, 45, 46, 48, 50, 52
Preparation . . . . . . . . . . . . . . . . . . 81, 82, 83, 84, 86
Preparation Proof test . . . . . . . . . . . . . 94, 95, 97, 99
Proof test . . . . . . . . . . . . . . . . . . . . . . . . . . . 94, 101
Safety mode . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
Simulation and check failure current . . . . . . . . . . . 99

X

XML build number (Parameter) . . . . . . . . . . . . . . . . 174

Z

Zero (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 144, 145
Zero adjustment (Parameter) . . . . . . . . . . . . . . . 42, 136
Zero adjustment offset (Parameter) . . . . . . . 87, 91, 137

Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Temperature unit (Parameter) . . . . . . . . . . . 39, 41, 131
Terminal current (high alarm) (Parameter) . . . . . . . 100
Terminal current (Parameter) . . . . . . . . . . . . . 135, 149
Terminal current 1 (Parameter) . . . . . . . . . . . . . . . . 101
Terminal current 2 (Parameter) . . . . . . . . . . . . . . . . 102
Terminal voltage 1 (Parameter) . . . . . . . . . . . . 116, 134
Tertiary variable (TV) (Parameter) . . . . . . . . . . . . . . 153
Timestamp (Parameter) . . . . . . . . . . . . . . . . . . 106, 107Timestamp Baseline (Parameter) . . . . . . . . . . . . . . . . 72
Timestamp previous Baseline (Parameter) . . . . . . . . . 73
Timestamp stored CRC device config. (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86, 97, 175
Tolerated deviation +/- (Parameter) . . . . . . . . . . 73, 115

U

Upper Range Limit (Parameter) . . . . . . . 44, 46, 48, 139
Upper range value output (Parameter). . . . . . . . . . . . . . . . 45, 49, 51, 90, 142, 145, 146, 148
Upper sensor trim (Parameter) . . . . . . . . . . . . . . . . 139
Upper sensor trim measured value (Parameter) . . . . 138
User management (Submenu) . . . . . . . . . . . . . . . . . 164
User role (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 164
User temperature process alert (Parameter) . . . . 79, 120

V

Value 1 display (Parameter) . . . . . . . . . . . . . . . . . . 167
Value 2 display (Parameter) . . . . . . . . . . . . . . . . . . 168
Value 3 display (Parameter) . . . . . . . . . . . . . . . . . . 168
Value 4 display (Parameter) . . . . . . . . . . . . . . . . . . 169
Value current output (Parameter) . . . . . . . . . . . . . . 113
Value pressure simulation (Parameter) . . . . . . . . . . 113
Verification result (Parameter) . . . . . . . . . . . . . . 59, 114
Visual inspection (Parameter) . . . . . . . . . . . . . . . . . . 95

W

Wet calibration (Submenu) . . . . . . . . . . . . . . . 144, 145
WizardCommissioning . . . . . . . . . . . . . . . . . . . . . . . . . . 34
Confirmation . . . . . . . . . . . . . . . . . . . 87, 88, 89, 91
Device identification . . . . . . . . . . . . . . . . . 34, 36, 37
Finish . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 93, 103
Heartbeat Verification . . . . . . . . . . . . . . . . . . . . . 55
Locking . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 91
180

Endress+Hauser

*71647907*
71647907

www.addresses.endress.com