https://kamemizu.co.jp

(let a patient rinse the mouth thoroughly).

20500BZZ00089000

And if the material sticks to the skin,

Ⅷ．CLINICAL PROCEDURES
1. Remove an old resin surface of adenture base to obtain a new surface

rinse with plenty of running water with

Short-term soft lining material for denture basesoap. If it sticks to clothing, immediately

using bur, and rinse with water and dry it.

Denture Soft Ⅱ

wipe it oﬀ with alcohol (with cotton ball)

2. Measure both powder and liquid. In case ofthoroughly.

lower jaw of a full denture, the standard
ratio of powder to liquid is the powder

INSTRUCTIONS FOR USE
Ⅰ．INTRODUCTION

２. Handling and manipulation precautions

adjusting the scale of 3 (3g) of the

1. When a mucous membrane becomesmeasuring cup to the liquid adjusting theBy lining mucous membrane face of ainﬂamed or there is a big diﬀerence inscale 2 (2mL) of a pipette. In case ofdenture base, it is used for temporary

the distribution of bite pressure against

upper jaw of a full denture, the quantity

lining,tissue condition, obtaining functionalthe mucous beneath the denture base,of powder and liquid is twice as much ofimpression and so on.

the mucous membrane needs treatment
before hand..

Ⅱ．INDICATIONS

what mentioned above.
3. Add the liquid to the powder in the

2. As the liquid is inﬂammable, it should

mixing cup with a pipette, and mix it for

Short-term soft lining material for denture

not be used or placed near the ﬁre. And

about 20 seconds not to involve air withbase,and for functional impression.

if it is spilt on the table or the ﬂoor,

a spatula. After 1 to 2 minutes thewipe it immediately with dry cloth.
Ⅲ．CONTRAINDICATIOＮ

3. It should be used in the place where themixture will become like syrup.
4. Spread the syrup like mixture onto theIt should not be used with those patients

ventilation is appropriate (several ventilation

inner side of the denture base, smooth it

who had allergic reactions including rash,

per hour).

even as much as possible with a spatulaskin inﬂammation to methyl methacrylate

4. When polymerized denture is ground or

polymer, methyl methacrylate monomer or

polished, use a local ventilation or dust

organic solvents in the past.

mask approved by public organizations
not to inhale ﬁne particles.

Ⅳ．POSSIBLE SIDE EFFECT

and insert it in a mouth.
5. Ask a patient for occlusion, and take out
the denture about 5 minutes later.
6. Take oﬀ any excess material with a knife

5. In order to prevent infection, syringes

While it is used, allergic reactions including
redness, sore, blister, swelling may occur.

and spatulas should be cleaned, andor a hot spatula.
7. Apply New top coat as a surface

sterilized with alcohol after use.

smoothing material thoroughly to the

6. It should not be used together with
Ⅴ．INCOMPATIBILITIES

lining surface and the part whereother products.

Do not use metal plates.

mucous surface meets periphery. And

7. It should not be used in any ways othersmooth uneven spots.

than resin for denture base repair.
Ⅵ．PRECAUTIONS

8. It should not be used by any person

１．Safety precautions

other than qualiﬁed dentists.

1. If allergic reactions including rash, eczema,
redness, tumor, swelling, itch, eruption,[CAUTION]
1. The mixture should not be spread too
much, as it will cause to ﬂow into the

３．Storage precautions

pharynx. Ask the patient to breath through1. The material should be stored to avoid

his nose so that it will not ﬂow into thematerial, stop using it immediately and let theﬁre, high temperature, high humidity

pharynx, which should be conﬁrmed in thepatient call a doctor, or call a doctor in caseand direct sunlight, at 0℃ to 23℃.

case of patients who has lower reﬂex

numbness occur to a patient due to using theof a dentist or a medical staﬀ suﬀered.

2. It should not be stored in a large quantity

2. When using the material, take care

(Such as aged people).

in the same storage. A ﬁre extinguisher

2. When the material is taken out from thesuﬃciently so as not to contact with eyes,

should be installed in the place where it iscontainer, use it immediately, and theskin, or clothing by wearing medical gloves

used and the storage.

container of the liquid should be tightenedand safety glasses. Although medical

3. As for both the powder and the liquid, lids ofgloves (for dentistry) prevent it fromthe containers should be tightened ﬁrmly.

contacting with hands directly, as it is4. While it is not used for a long period ofits lid ﬁrmly after use.
[WARRANTY]

known, organic solvents or monomers

time, or stored laying them on their side,

KAMEMIZU CHEMICAL IND.CO., LTD. will

penetrate them in a short time. So, if thestore it with both a lid and an inner

replace any product that is proved to beliquid, the powder, the mixture, or thestopper tightened. (Do not throw away

defective. KAMEMIZU CHEMICAL IND.CO.,

bonding material sticks to gloves, throw

the inner stopper and keep it for storage.)LTD. does not accept liability for any loss

them away immediately, and rinse hands

5. The material should be properly stored

thoroughly with running water with soap.

and controlled, and not be touched byarising out of the application or use of or

any person other than people engage in

the inability to use these products. Before

dentistry

using, the user shall determine the3. Take suﬃcient precautions so that thematerial will not contact with patientseyes, skin, or clothing. In addition, try not
to let it stick to any other parts of mucous
in the mouth as much as possible than thepart that it should be applied.
4. If it contacts with patient s eyes, rinse themwith plenty of running water immediately, andlet him call an eye doctor, or call an eye
doctor in case of a dentist or a medical
staﬀ suﬀered.
5. If excess resin ﬂows into the respiratory

suitability of the products for the intended
Ⅶ．COMPONENTS
１．Powder100gliability whatsoever in connection therewith.

and others
２．Liquid80mL

KAMEMIZU CHEMICAL IND.CO.,LTD.

BPRG, Ethyl alcohol, and others

17-16 TOYOSATO-CHO, NEYAGAWA,

３．New top coat 6mL

OSAKA 572-0071, JAPAN

Ethyl acetate, and others
４．Accessories
Powder measuring cup, Mixing cup,

a medical specialist and have a treatment.

Pipette, Spatula, Steeple-crowned cap

immediately clean the mouth suﬃciently

use and the user assumes all risk andPoly(ethyl methacrylate), coloring agent,

tract or the esophagus, let a patient call
6. After having ﬁnished treatment in the mouth,or damage, direct, consequential or special,ECREP

Emergo Europe
Molenstraat 152513 BH The Hague
The Netherlands
Tel : (31) (0) 70 345-8570
Fax: (31) (0) 70 346-7299