GSD NovaType Select L452R
SARS-CoV-2
RT-PCRRUO

FOR RESEARCH USE ONLY
Not for use in diagnostic procedures

English ..........................................................................................................................................................................2
Deutsch .........................................................................................................................................................................7
Bibliography / Literatur / Bibliographie / Bibliografia / Bibliografía / Bibliografia .........................................................15
Abbreviations / Abkürzungen / Abréviations / Abbreviazioni / Abreviaciónes / Abreviaturas .....................................15
Symbols Key / Symbolschlüssel / Explication des Symboles / Legenda / Símbolos / Tabela de símbolos ...............16

Product Number:

PCOVL452R3PCOVL452R4(1 x 96 Determinations)(4 x 96 Determinations)

ENGLISH
1. INTRODUCTION
End of 2019, a novel respiratory disease emerged in the city of Wuhan, Hubei Province of the People’s Republic of China, andsoon spread rapidly within the country and worldwide. The causative agent was identified as severe acute respiratory syndrome
coronavirus 2 (SARS-CoV-2). SARS-CoV-2 (2019-nCoV), like the closely related SARS coronavirus (SARS-CoV), belongs to thegenus Betacoronavirus within the family of coronaviruses. The zoonotic reservoir of the virus appears to be bats.
Coronaviruses are enveloped, positive single‐stranded large RNA viruses that infect humans, but also a wide range of animals.
The common human coronaviruses NL63, 229E, OC43 and HKU1 are widespread especially throughout the winter months. They
are responsible for up to one third of all acute respiratory diseases, typically with mild symptoms (common cold). More than 80 %of the adult population have antibodies against human coronaviruses. The immunity from previous infections lasts only for a shortperiod of time. Therefore, reinfections with the same pathogen are possible just after one year.
SARS-CoV-2 is predominantly transmitted by droplet infection via coughing or sneezing and through close contact with infectedpatients. In theory, smear infection and infection through the conjunctiva of the eyes are also possible.
The incubation period is in the median 5–6 days (and up to 14 days maximum).
The clinical manifestations of SARS‐CoV‐2‐related COVID-19 disease include fever, cough, respiratory problems and fatigue. In
most patients the infection manifests with symptoms of a mild febrile illness with irregular lung infiltrates.
The initial clinical sign of COVID‐19 which allowed case detection was pneumonia. But it turned out that the course of the disease
is non-specific and varies widely, from asymptomatic courses to severe pneumonia with lung failure and death. However, basedon current knowledge, around 80 % of the illnesses are mild to moderate.
Although severe courses of the disease also occur in younger patients and people without previous illness, the following groupsof people have an increased risk of serious forms of the disease: elderly people (with a steadily increasing risk from around 50-60
years of age), smokers and people with certain diseases of the cardiovascular system or the lungs, patients with chronic liverdiseases, diabetes mellitus, cancer, or patients with a weakened immune system (e.g. due to immune deficiencies or by taking
drugs that suppress the immune system).
Treatment focuses on supportive measures depending on the severity of the clinical picture (e.g. oxygen administration, fluidbalance management, etc.) as well as the treatment of relevant underlying diseases. Various specific therapeutic approaches(directly antiviral effective, immunomodulatory effective) have been and are being investigated in studies during the course of the
SARS-CoV-2 pandemic.
The SARS-CoV-2 trimeric spike glycoprotein (S) is exposed on the surface of virions and mediates viral entry into host cells. The
S protein constitutes the primary target of all current leading vaccine candidates. This large, exposed protein is readily targetedby neutralizing antibodies, which indirectly creates selective pressure for the emergence of evasion mutations. The propensity for
S to mutate may limit its future use in serological assays and vaccines, as antibodies directed against the current variant may notbind emerging mutated epitopes.
Gene mutations in emerging SARS-CoV-2 variants can affect transmission, virulence, and immunity. Variants for which there isevidence (variants of concern, VOC) or suspicion (variants of interest, VOI) of increased transmissibility, more severe disease(e.g., increased hospitalizations or deaths), significant reduction in neutralization by antibodies, reduced efficacy of treatments or
vaccines, or diagnostic detection errors are considered to be under special surveillance worldwide. Current VOCs and VOIs
include the lineages, B.1.1.7 (501Y.V1), B.1.351 (20H/501Y.V2), B.1.1.28 (P.1; 20J/501Y.V3), B.1.617 (G/452.V3) andB.1.427/B.1.429 (CAL.20C). They all exhibit mutations in the spike protein that are associated with increased transmissibility
and/or suspected to negatively affect vaccine efficacy by avoiding the immune response. The lineages B.1.617 (G/452.V3) andB.1.427/B.1.429 (CAL.20C) are both found to be carrier of the mutation L452R in addition to other mutations. This mutation issuspected to be associated with both increased infectivity and reduced immune response and is therefore of relevance to theglobal pandemic situation.
Species
SARS-CoV-2(severe acuterespiratory syndrome
coronavirus 2)

DiseaseSymptoms e.g.

Transmission route

COVID-19

the course of the disease is unspecific,
diverse and varies greatly, fromasymptomatic courses to severe pneumonia
with lung failure and death

primary mode of transmission: droplet
infection;
smear infections and infections via theconjunctiva of the eyes aretheoretically possible

Infection or presence of pathogen may be identified by▪
Nucleic acid Amplification Techniques (NAT): e.g. RT-PCR▪
Serology:
detection of antibodies by e.g. ELISA

2. INTENDED USE (FOR RESEARCH USE ONLY!)
The GSD NovaType Select L452R SARS-CoV-2 is a research use only (RUO) real-time reverse transcription PCR (RT-PCR)
assay designed for the qualitative detection of the SARS-CoV-2 (Severe acute respiratory syndrome coronavirus 2) S genemutation L452R. Sample material should be genomic RNA extracted from human upper respiratory (nasal wash/smear,
nasopharyngeal wash/smear, oropharyngeal swab) specimen types that have previously tested positive for SARS-CoV-2 RNA by
diagnostic RT-PCR methods e.g. GSD NovaPrime® SARS-CoV-2 (COVID-19). The GSD NovaType Select L452R SARS-CoV-2
is expressly not intended for diagnosis of SARS-CoV-2 infection and not for use in diagnostic procedures. It is only designed toaid investigations related to the prevalence and frequency of virus variants.

2

3. PRINCIPLE OF THE ASSAY
The qualitative determination of specific RNA is based on Real-Time reverse-transcription Polymerase Chain Reaction (RT-PCR)
technology. The kit contains specific primers and probes labelled with fluorescent reporter and quencher molecules for
amplification and simultaneous detection of the specific RNA sequence representing the SARS-CoV-2 S gene mutation L452R.
The S gene specific probe is labelled with the fluorophore FAMTM. As internal process control, the human Ribonuclease P(RNase P) is detected using an CY5TM labelled probe. The human RNase P gene primer and probe set serves as an internal
positive control to monitor sample quality, RNA extraction, and for detection of inhibitors of the PCR reaction.
The GSD NovaType Select L452R SARS-CoV-2 has been validated for following real-time PCR platforms:▪
AriaMxTM (Agilent Technologies)▪
AriaDxTM (Agilent Technologies)

4. MATERIALS
4.1.

Reagents Provided
Cap

SymbolComponent

green

E-MIX

(reverse transcriptase, hot-start DNA polymerase, RNase inhibitor,
nucleotides, magnesium, stabilizers and buffer)

blue/white

PP│L452R

red

PC

transparent

NFW

PCOVL452R3 (PCOVL452R4)
Volume per Vial
Number of
[µL]
Vials

RT-PCR Enzyme Mix
Primer-Probe-Mix(for detection of mutation L452R and for detection of RNase P gene)

Positive Control(plasmid DNA representing SARS-CoV-2 S gene mutation L452R)

Nuclease Free Water

550

1 (4)

350

1 (4)

150

1 (4)

500

1 (4)

4.2.

Materials and Equipment needed, but not provided▪
▪

Deionized nuclease-free and nucleic-acid free water
Real-Time PCR instrument, equipped with suitable set of optical cartridges (for already validated instruments refer to 3.
PRINCIPLE OF THE ASSAY; for dyes used refer to 7.2.)
Alternative Real-Time PCR instruments might also be appropriate. Their suitability for use with the GSD NovaType Select
L452R SARS-CoV-2 has to be validated by the user.
Appropriate Real-Time PCR consumables (e.g. disposable tubes, reaction plates, corresponding optical closing
materials)
Benchtop microcentrifuge
Centrifuge with a rotor for microtiter plates
Vortex mixer
Adjustable pipettes in relation to reaction setup
Disposable DNase/RNase free pipette tips with filters
Disposable powder-free gloves

▪
▪
▪
▪
▪
▪
▪

5. STABILITY AND STORAGE
The GSD NovaType Select L452R SARS-CoV-2 kit is shipped on dry ice and all components should arrive frozen.
▪
All components have to be stored between -30 °C and -15 °C immediately after arrival.
▪
Repeated freeze thaw cycles (more than two) of reagents should be avoided, since this might affect the performance of
the kit. Reagents should be frozen in aliquots if they are used intermittently.
▪
Keep unfrozen storage (e.g. storage on ice) as short as possible.
▪
Protect E-MIX and PP│L452R from light and keep frozen until use.

6. SAMPLE PREPARATION▪

▪

Extracted RNA or total nucleic acid extracted from human respiratory specimen types (nasal lavage/smear,
nasopharyngeal lavage/smear, oropharyngeal swab, bronchoalveolar lavage) previously tested positive for SARS-CoV-2
RNA by RT-PCR methods e.g. GSD NovaPrime® SARS-CoV-2 (COVID-19) is the starting material for the GSD
NovaType Select L452R SARS-CoV-2.
The quality and the quantity of the extracted RNA has a crucial effect on the performance of the entire RT-PCR test
system. Make sure that the nucleic acid extraction method is compatible with Real-Time PCR technology.

3

7. ASSAY PROCEDURE
7.1.

Reaction Setup▪

Please read the instructions for use carefully before performing the assay. Reliability of results depends on following
strictly the instructions for use.
Before use make sure that all samples and reagents are thawed completely, mixed by up and down pipetting or vortexing
and centrifuged briefly.
Important:
E-MIX is viscous. Briefly spin down the tube to ensure material has not lodged in the cap or side of
tube. Be sure to pipette and dispense carefully, and use pipette tips suitable for pipetting viscous liquids.
The use of NFW as no template control (NTC) is highly recommended.
Define the positions (wells) for samples and controls ( PC or NFW ) on the plate.

▪
▪
▪
▪

Reaction Setup
Component

Volume

E-MIX

5 μL

PP│L452R

3 µL

Sample or PC or NFW

12 μL

Total Volume

20 μL

▪
▪

Close the optical reaction plate with corresponding optical closing material.
Centrifuge the optical reaction plate in a centrifuge with a rotor for microtiter plates for 60 seconds at approximately
1000 x g (~ 3000 rpm).

7.2.

Programming the Real-Time PCR Instrument

Regarding setup and programming of the Real-Time PCR instrument, please use the manual of the respective instrument.
For detailed programming instructions regarding the use of the GSD NovaType Select L452R SARS-CoV-2 on specific Real-Time
PCR instruments please contact NovaTec Immundiagnostica GmbH.
RT-PCR Run Settings
Reaction Volume

20 μL

Passive Reference

None

Fluorescent Detectors / Fluorophores
Detection of the amplified viral nucleic acid fragment for the S gene mutation L425R is performed in the FAMTM detection channel
with MGBEQ Quencher. Detection of the amplified internal process control RNase P is performed in the detection channel CY5TM
with BHQ2 Quencher.
TargetFluorophore/Detection Channel

QuencherS gene L452R

FAMTM

MGBEQ

RNAse P

CY5TM

BHQ2

Temperature Profile and Data Collection
No. of Cycles

Temperature

Time (min)Data Collection

1

50 °C

10:00

-

1

95 °C

03:00

-

95 °C

00:10

60 °C

00:30

Fluorescence measurement at
the end of every cycle

40

Before starting the test run, please check the settings for cycles, temperature and time.

48. RESULTS
Data analysis should be performed with the software of the used real-time PCR device according to manufacturer’s instructions.
Analysis settings:
Setting

Recommendation

ThresholdEnter a value for the threshold so that the threshold is:• Above the background• Below the plateau and linear regions of the amplification curve• Within the exponential phase of the amplification curve

Baseline

Select start and end cycle values so that possible initial signal noise is not included and the baseline ends
before significant fluorescence is detected.

8.1.

Interpretation of Results

The test run is valid only if the RT-PCR run is complete.
The GSD NovaType Select L452R SARS-CoV-2 test protocol dictates that the controls be analyzed before sample results.

8.1.1.

Controls

If kit control(s) fail, the test is invalid and needs to be repeated.
Sample data is analyzed and interpreted only after all the kit controls pass.
▪

NTC (no template control):must NOT have a detectable Ct in the S gene or RNase P reactions. If this control has adetectable Ct in any of the reaction wells, this indicates contamination of the PCR run and it is considered invalid andmust be repeated.
PC :Ct ≤ 38 for S gene L452R target (FAMTM)Internal process control (RNase P): all human samples should exhibit Ct values Ct ≤ 38, thus indicating the presenceof the human RNase P gene.

▪
▪

Failure to detect RNase P in any specimens may indicate:▪
Improper extraction of nucleic acid from clinical materials resulting in loss of RNA and/or RNA degradation.
▪
(RT-) PCR inhibition.
▪
Absence of sufficient human cellular material due to poor collection or loss of specimen integrity.
▪
Improper assay set up and execution.
▪
Reagent or equipment malfunction.
If the RNase P assay does not produce a positive result for specimens, interpret as follows:▪
If S gene marker is positive even in the absence of a positive RNase P, the result should be considered valid. It is possiblethat some samples may fail to exhibit RNase P amplification curves due to low copy numbers in the original sample. A
negative RNase P signal does not preclude the presence of SARS-CoV-2 RNA in a specimen.
▪
If the S gene marker AND RNase P are negative for the specimen, the result should be considered invalid for the
specimen. If residual specimen is available, repeat the extraction procedure and repeat the test. If all markers remain
negative after re-test, report the results as invalid and a new specimen should be collected if possible.

8.1.2.

SamplesIf test run is valid interpretation of sample results is as follows:▪
Positive (POS): Ct ≤ 38▪
Negative (neg): Not detected (ND) or Ct > 38
The probe for detection of the S gene L452R mutation is labelled with FAMTM. The probe for detection of the RNase P gene islabelled with CY5TM. Please refer to the following table for interpretation of different signal combinations.
Detection Channel
FAMTM (L452R)

CY5TM (RNAse P)

Positive

Positive or Negative

Negative

Positive

Negative

Negative

Interpretation of Results
S gene L452R mutation detected
S gene L452R mutation not detected.
This result is an indication of the presence of the L452 wildtype sequence dueto the positive result in the preceding SARS-CoV-2 RT-PCR screening assay
PCR inhibition or reagent failure.
A diagnostic statement must not be made.
The RT-PCR run should be repeated or a new sample must be analyzed.

9. LIMITATIONS OF THE PROCEDURE
Very low amounts of starting material, e.g. weak positive samples (high Ct values in the previous diagnostic SARS-CoV-2
RT-PCR), can result in GSD NovaType Select L452R SARS-CoV-2 test runs with negative signals. However, such results do not
exclude the presence of SARS-CoV-2 RNA and therefore do not exclude the presence of a L452R mutation.

510. TRADEMARKS AND DISCLAIMERS
Registered names, trademarks, etc. used in this document are to be considered protected by law even if not specifically marked
as such.

11. PRECAUTIONS AND WARNINGS
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪

▪
▪
▪
▪

FOR RESEARCH USE ONLY. Not for use in diagnostic procedures.
All human samples should be regarded and handled as potentially infectious.
Do not interchange reagents of different production lots.
No reagents of other manufacturers should be used along with reagents of this test kit.
Do not use reagents after expiry date stated on the label.
Wear disposable powder-free gloves, a laboratory coat and eye protection when handling specimens.
Always use DNase/RNase-free disposable reaction tubes and pipette tips with aerosol barriers.
Avoid microbial and nuclease (DNase/RNase) contamination of the specimen and the components of the kit.
In order to avoid contamination of working space with nucleic acids, reaction tubes/plates should not be opened after
amplification.
RT-PCR is highly sensitive to nucleic acid contamination. Therefore, positive / potentially positive material must be stored
separately from all other components of the kit.
Dedicate supplies and equipment to the separate working areas and do not move them from one area to another.
This assay must not be used on the specimen directly.
Prior to using this assay the nucleic acid has to be extracted with suitable extraction methods from the original specimen.
Since ethanol is a strong Real-Time PCR inhibitor, it is necessary to completely eliminate it prior to the elution of thenucleic acid during extraction. If using spin columns with washing buffers containing ethanol, it is highly recommended
to perform an additional centrifugation step of 10 min at approximately 17000 x g (~ 13000 rpm) before eluting the RNA.
For this additional centrifugation step, use a new collection tube.
The result of this RT-PCR kit may be influenced by potential mutations in the genome of the pathogen if they are locatedin the primer / probe binding region. Underestimation and/or failure to detect the pathogen may occur.
RT-PCR inhibitors may also elicit underestimation, false negative results or invalid runs. Therefore, only use nucleic acids
extraction kits, which remove RT-PCR inhibitors and which are dedicated for downstream RT-PCR processes.
The Real-Time PCR is only designed for qualified personnel who are familiar with good laboratory practice and trainedin RT-PCR.
For further internal quality control each laboratory should additionally use known samples.

11.1. Disposal Considerations
Residues of chemicals and preparations are generally considered as hazardous waste. The disposal of this kind of waste isregulated through national and regional laws and regulations. Contact your local authorities or waste management companies
which will give advice on how to dispose hazardous waste.

12. ORDERING INFORMATION
Prod. No.:PCOVL452R3PCOVL452R4

GSD NovaType Select L452R SARS-CoV-2
GSD NovaType Select L452R SARS-CoV-2

6(1 x 96 Determinations)(4 x 96 Determinations)DEUTSCH
1. EINLEITUNG
Ende 2019 trat in der Stadt Wuhan, Provinz Hubei, Volksrepublik China, eine neuartige Atemwegserkrankung auf, die sich schonbald innerhalb des Landes und rasch weltweit ausbreitete. Als Erreger wurde das SARS-CoV-2 (“Schweres akutes
Atemwegssyndrom Coronavirus 2”) identifiziert. SARS-CoV-2 (2019-nCoV) gehört, wie das eng verwandte SARS-Coronavirus(SARS-CoV), zur Gattung der Betacoronaviren innerhalb der Familie der Coronaviren. Das zoonotische Reservoir des Virus sindvermutlich Fledermäuse.
Coronaviren sind große, von einer Lipidhülle umgebene RNA-Viren mit einzelsträngigem Plus-Strang-Genom, die den Menschen,aber auch eine Vielzahl von Tieren infizieren. Die bekannten humanen Coronaviren NL63, 229E, OC43 und HKU1 sind vor allemin den Wintermonaten weit verbreitet. Bis zu einem Drittel aller akuten Atemwegserkrankungen, typischerweise mit milden
Symptomen (Erkältung), werden durch diese Coronaviren verursacht. Bei mehr als 80 % der erwachsenen Bevölkerung lassensich Antikörper gegen humane Coronaviren nachweisen. Die Immunität gegenüber früheren Infektionen hält nur für eine kurze
Zeit an. Reinfektionen mit dem gleichen Erreger sind daher bereits nach einem Jahr möglich.
SARS-CoV-2 wird vorwiegend durch Tröpfcheninfektion über Husten oder Niesen und durch engen Kontakt mit infizierten
Personen übertragen. Theoretisch sind auch Schmierinfektionen und Infektionen über die Bindehaut der Augen möglich.
Die Inkubationszeit liegt im Median bei 5-6 Tagen (und maximal bei bis zu 14 Tagen).
Zu den klinischen Manifestationen der SARS-CoV-2-assoziierten COVID-19 Erkrankung zählen Fieber, Husten,
Atembeschwerden und Müdigkeit. Bei den meisten Patienten manifestiert sich die Infektion mit Symptomen einer leichten
fieberhaften Erkrankung mit irregulären Lungeninfiltraten.
Das ursprüngliche klinische Anzeichen von COVID-19, das eine Diagnose ermöglichte, war eine Lungenentzündung. Es stelltesich jedoch heraus, dass der Krankheitsverlauf unspezifisch ist und stark variiert: von asymptomatischen Verläufen bis hin zu
einer schweren Lungenentzündung mit Lungenversagen und Tod. Nach heutigem Kenntnisstand sind jedoch etwa 80 % der
Erkrankungen leicht bis moderat.
Obwohl schwere Krankheitsverläufe auch bei jüngeren Patienten und Menschen ohne Vorerkrankungen auftreten, haben
folgende Personengruppen ein erhöhtes Risiko für schwere Erkrankungsformen: ältere Menschen (mit einem stetig steigenden
Risiko ab etwa 50-60 Jahren), Raucher und Menschen mit bestimmten Erkrankungen des Herz-Kreislauf-Systems oder der
Lunge, Patienten mit chronischen Lebererkrankungen, Diabetes mellitus, Krebs oder Patienten mit einem geschwächten
Immunsystem (z.B. durch Immunschwäche oder durch die Einnahme von Medikamenten, die das Immunsystem unterdrücken).
Die Behandlung beschränkt sich auf unterstützende Maßnahmen in Abhängigkeit vom Schweregrad des Krankheitsbildes (z.B.
Sauerstoffgabe, Flüssigkeitsmanagement etc.) sowie auf die Behandlung relevanter Grunderkrankungen. Verschiedene
spezifische Therapieansätze (direkt antiviral, immunmodulatorisch wirksam) wurden und werden im Verlauf der SARS-CoV-2Pandemie im Rahmen von Studien untersucht.
Das trimere Spike-Glykoprotein (S) ist auf der Oberfläche der Virionen exponiert und vermittelt den Eintritt des Virus in die
Wirtszellen. Alle derzeit führenden Impfstoffkandidaten konzentrieren sich auf das Spike-Protein als Zielantigen. Dieses große,
exponierte Protein kann gut von neutralisierenden Antikörpern erkannt werden, was indirekt einen Selektionsdruck für das
Auftreten von Ausweichmutationen erzeugt. Die Tendenz S-Proteins zu mutieren, könnte seine künftige Verwendung inserologischen Tests und Impfstoffen einschränken, da Antikörper, die gegen die aktuelle Variante gerichtet sind, neu entstehendemutierte Epitope möglicherweise nicht binden.
Genmutationen in neu auftretenden SARS-CoV-2-Varianten können sich auf die Übertragung, Virulenz und Immunität auswirken.
Varianten, für die es Nachweise (variants of concern, VOC) oder Verdachtsmomente (variants of interest, VOI) für eine erhöhteÜbertragbarkeit, eine schwerere Erkrankung (z.B. vermehrte Krankenhausaufenthalte oder Todesfälle), eine signifikante
Verringerung der neutralisierenden Wirkung von Antikörpern, eine verminderte Wirksamkeit von Behandlungsmethoden oderImpfstoffen gibt oder es zu Fehlern bei der Diagnostik kommt, gelten weltweit als besonders überwachungsbedürftig. Aktuelle
VOCs und VOIs umfassen die Linien B.1.1.7 (501Y.V1), B.1.351 (20H/501Y.V2), B.1.1.28 (P.1; 20J/501Y.V3), B.1.617 (G/452.V3)
und B.1.427/B.1.429 (CAL.20C). Sie alle weisen Mutationen im Spike-Protein auf, die mit einer erhöhten Übertragbarkeit
assoziiert sind und/oder im Verdacht stehen, die Wirksamkeit des Impfstoffs durch Umgehung der Immunantwort negativ zu
beeinflussen. Die Linien B.1.617 (G/452.V3) und B.1.427/B.1.429 (CAL.20C) sind neben anderen Mutationen auch Träger der
Mutation L452R. Es wird vermutet, dass diese Mutation sowohl mit einer erhöhten Infektiosität als auch mit einer verminderten
Immunantwort einhergeht und daher für die globale Pandemiesituation von Bedeutung ist.
Spezies
SARS-CoV-2(“Schweres akutes
Atemwegssyndrom
Coronavirus 2”)

Erkrankung

Symptome (z.B.)

COVID-19

Der Krankheitsverlauf ist unspezifisch, vielfältig
und variiert stark, von asymptomatischen
Verläufen bis hin zu schwerer Lungenentzündung
mit Lungenversagen und Tod

Nachweis des Erregers bzw. der Infektion durch:▪
Nukleinsäure-Amplifikations-Technik (NAT):▪
Serologie:

Infektionsweg
Primärer Übertragungsweg:
Tröpfcheninfektion;
Schmierinfektionen und
Infektionen über die Bindehaut derAugen sind theoretisch möglichz.B. RT-PCR
Nachweis von Antikörpern durch z.B. ELISA

7

2. VERWENDUNGSZWECK (NUR FÜR FORSCHUNGSZWECKE!)Die GSD NovaType Select L452R SARS-CoV-2 ist ein Real-Time Reverse-Transkriptions-Polymerase-Kettenreaktions-Assay(RT-PCR) nur für Forschungszwecke (Research Use Only, RUO), der für den qualitativen Nachweis der SARS-CoV-2 („Schweres
Akutes Atemwegssyndrom Coronavirus 2“) S-Gen Mutation L452R entwickelt wurde. Als Probenmaterial sollte genomische RNA
verwendet werden, die aus humanen respiratorischen Proben (Nasenspülung/-abstrich, Nasopharyngealspülung/-abstrich,Oropharyngealabstrich) extrahiert wurde und die zuvor mittels diagnostischer RT-PCR-Methoden, z.B. der GSD NovaPrime®
SARS-CoV-2 (COVID-19) positiv auf SARS-CoV-2 RNA getestet wurde. Die GSD NovaType Select L452R SARS-CoV-2 ist
ausdrücklich nicht für die Diagnose einer SARS-CoV-2-Infektion und nicht für den Einsatz in diagnostischen Verfahren
vorgesehen. Sie soll lediglich Untersuchungen zur Prävalenz und Häufigkeit von Virusvarianten unterstützen.

3. TESTPRINZIP
Die qualitative Bestimmung spezifischer RNA basiert auf der Real-Time Reverse-Transkriptions-Polymerase-KettenreaktionsTechnologie (RT-PCR). Der Kit enthält spezifische Primer und Sonden, die mit fluoreszierenden Reporterfarbstoffen und
Quenchern zur Amplifikation und zum gleichzeitigen Nachweis der spezifischen RNA-Sequenz, welche die SARS-CoV-2 S GenMutation L452R repräsentiert, markiert sind. Die für das S-Gen spezifische Sonde ist mit den Fluorophor FAMTM markiert. Alsinterne Prozesskontrolle wird das humane Ribonuclease P (RNAse P) Gen mit einer CY5TM-markierten Sonde nachgewiesen.
Das Primer- und Sonden-Set für das humane RNase P-Gen dient als interne Positivkontrolle zur Überwachung derProbenqualität, der RNA-Extraktion und zum Nachweis von Inhibitoren der PCR-Reaktion.
Die GSD NovaType Select L452R SARS-CoV-2 wurde für folgende Real-Time PCR Plattformen validiert:▪
AriaMxTM (Agilent Technologies)▪
AriaDxTM (Agilent Technologies)

4. MATERIALIEN
4.1.

Mitgelieferte Reagenzien

DeckelSymbolKomponente

grün

E-MIX

(Reverse Transkriptase, Hot-Start DNA Polymerase, Rnase-Inhibitor,
Nukleotide, Magnesium and Stabilistoren in Puffer)

blau/weiß

PP│L452R

rot

PC

transparent

NFW

PCOVL452R3 (PCOVL452R4)
Volumen pro
AnzahlRöhrchen [µL]
Röhrchen

RT-PCR Enzym-Mix
550

1 (4)

350

1 (4)

(Plasmid-DNA, die die SARS-CoV-2 S Gen-Mutation L452R
repräsentiert)

150

1 (4)

Nukleasefreies Wasser

500

1 (4)

Primer-Probe-Mix(zum Nachweis der Mutation L452R und des RNase P Gens)

Positive Control

4.2.

Erforderliche Materialien und Geräte, nicht mitgeliefert▪
▪

Entionisiertes Nuklease- und Nukleinsäure-freies Wasser
Real-Time PCR Gerät, ausgestattet mit geeigneter Kombination an Filtern (für bereits validierte Geräte siehe 3.
TESTPRINZIP; für verwendete Fluorophore siehe 7.2.)
Alternative Real-Time PCR Geräte sind möglicherweise ebenfalls geeignet; ihre Eignung für die Verwendung mit der
GSD NovaType Select L452R SARS-CoV-2 muss vom Anwender validiert werden.
Geeignete Real-Time PCR-Verbrauchsmaterialien (z.B. Plastikröhrchen für den einmaligen Gebrauch, Reaktionsplatten,
entsprechende optische Verschlussmaterialien)
Tisch-Mikrozentrifuge
Zentrifuge mit Rotor für Mikrotiterplatten
Vortex-Mischer
Einstellbare Pipetten abhängig vom Reaktions-Setup
DNase/RNase-freie Einweg-Pipettenspitzen mit FilternPuderfreie Einweghandschuhe

▪
▪
▪
▪
▪
▪
▪

5. STABILITÄT UND LAGERUNG
Der GSD NovaType Select L452R SARS-CoV-2 Kit wird auf Trockeneis versandt und alle Komponenten sollten in gefrorenem
Zustand ankommen.
▪
Alle Komponenten müssen unmittelbar nach Ankunft zwischen -30 °C und -15 °C gelagert werden.
▪
Wiederholte Einfrier-Auftauzyklen (mehr als zwei) der Reagenzien sollten vermieden werden, da dies die Leistung desKits beeinträchtigen könnte. Reagenzien sollten aliquotiert eingefroren werden, wenn sie in Abständen verwendet
werden.
▪
Lagerung in aufgetautem Zustand (z.B. Lagerung auf Eis) so kurz wie möglich halten.
▪
E-MIX und PP│L452R vor Licht schützen und bis zur Verwendung tiefgekühlt aufbewahren.

8

6. VORBEREITUNG DER PROBEN▪

Extrahierte RNA oder Gesamtnukleinsäure aus humanen respiratorischen Proben (Nasenspülung/-abstrich,Nasopharyngealspülung/-abstrich, Oropharyngealabstrich, bronchoalveoläre Lavage), die zuvor mittels RT-PCRMethoden z.B. der GSD NovaPrime® SARS-CoV-2 (COVID-19) positiv auf SARS-CoV-2 RNA getestet wurde, ist das
Ausgangsmaterial für die GSD NovaType Select L452R SARS-CoV-2.
Die Qualität und Quantität der extrahierten RNA hat einen entscheidenden Einfluss auf die Leistung des gesamten RTPCR-Testsystems. Stellen Sie sicher, dass die Nukleinsäureextraktionsmethode mit der Real-Time PCR-Technologie
kompatibel ist.

▪

7. TESTDURCHFÜHRUNG
7.1.

Reaktions-Setup▪

Bitte lesen Sie die Gebrauchsanweisung vor der Durchführung des Assays sorgfältig durch. Die Zuverlässigkeit derErgebnisse hängt von der strikten Befolgung der Gebrauchsanweisung ab.
Stellen Sie vor Gebrauch sicher, dass alle Proben und Reagenzien vollständig aufgetaut, durch Auf- und Abpipettierenoder Vortexen gemischt und kurz zentrifugiert worden sind.
Wichtig:
Der E-MIX ist zähflüssig. Zentrifugieren Sie das Röhrchen kurz ab, um sicherzustellen, dass sich keinMaterial im Deckel oder an der Seite des Röhrchens abgesetzt hat. Achten Sie darauf, vorsichtig zu pipettieren und zu
dispensieren, und verwenden Sie Pipettenspitzen, die für das Pipettieren viskoser Flüssigkeiten geeignet sind.
Die Verwendung von NFW als „No Template Control“ (NTC) wird dringend empfohlen.
Definieren Sie die Positionen (Vertiefungen) für Proben und Kontrollen ( PC oder NFW ) auf der Platte.

▪
▪

▪
▪

Reaktions-Setup
Komponente

Volumen

E-MIX

5 μL

PP│L452R

3 µL

Probe oder PC oder NFW

12 μL

Gesamtvolumen

20 μL

▪
▪

Verschließen Sie die optische Reaktionsplatte mit dem entsprechenden optischen Verschlussmaterial.
Zentrifugieren Sie die optische Reaktionsplatte in einer Zentrifuge mit einem Rotor für Mikrotiterplatten für 60 Sekunden
bei etwa 1000 x g (~ 3000 U/min).

7.2.

Programmierung des Real-Time PCR-Geräts

Für die Konfiguration und Programmierung des Real-Time PCR-Geräts nehmen Sie bitte das jeweilige Handbuch zu Hilfe.
Für detaillierte Anweisungen zur Programmierung spezifischer Real-Time PCR-Geräte zur Verwendung der GSD NovaType
Select L452R SARS-CoV-2, wenden Sie sich bitte an NovaTec Immundiagnostica GmbH.
Einstellungen RT-PCR Lauf
Reaktionsvolumen

20 μL

Passive Referenz

Keine

Fluoreszenz-Detektoren / Fluorophore
Die Detektion des amplifizierten viralen Nukleinsäurefragments für die S Gen-Mutation L452R erfolgt im FAMTM Detektionskanal
mit MGBEQ Quencher. Die Detektion der amplifizierten internen Prozesskontrolle RNase P erfolgt im Detektionskanal CY5TM mit
BHQ2 Quencher.
TargetFluorophor/Detektionskanal

Quencher

S Gen L452R

FAMTM

MGBEQ

RNAse P

CY5TM

BHQ2

Temperaturprofil und Datenerfassung
Anzahl Zyklen

Temperatur

Zeit (min)Datenerfassung

1

50 °C

10:00

-

1

95 °C

03:00

-

95 °C

00:10

60 °C

00:30

Fluoreszenzmessung am
Ende jedes Zyklus

40Überprüfen Sie bitte die Einstellungen für Zyklen, Temperatur und Zeit bevor Sie den Testlauf starten.

9

8. ERGEBNISSE
Die Datenanalyse sollte mit der Software des verwendeten Real-Time PCR-Gerätes gemäß den Anweisungen des Herstellers
durchgeführt werden.
Analyse-Einstellungen:Einstellung

Schwellenwert

Basislinie

8.1.

Empfehlung
Geben Sie einen Wert für den Schwellenwert ein, so dass für den Schwellenwert gilt:• Oberhalb des Hintergrunds• Unterhalb des Plateaus und der linearen Bereiche der Amplifikationskurve• Innerhalb der exponentiellen Phase der Amplifikationskurve
Wählen Sie Start- und Endzykluswerte so, dass ein mögliches anfängliches Signalrauschen nicht
eingerechnet wird, und die Basislinie endet, bevor eine signifikante Fluoreszenz detektiert wird.

Interpretation der Ergebnisse

Der Testlauf ist nur gültig, wenn der RT-PCR Testlauf abgeschlossen ist.
Das GSD NovaType Select L452R SARS-CoV-2 Testprotokoll schreibt vor, dass die Kontrollen vor den Ergebnissen der Proben
analysiert werden müssen.

8.1.1.

Kontrollen

Wenn die Kit-Kontrolle(n) versagen, ist der Test ungültig und muss wiederholt werden.
Daten von Proben dürfen erst dann analysiert und interpretiert werden, wenn alle Kit-Kontrollen valide Ergebnisse zeigen.
▪

NTC (no template control):darf KEINEN nachweisbaren Ct in den S Gen- oder RNase P-Reaktionen aufweisen.
Wenn diese Kontrolle in einer der Vertiefungen einen nachweisbaren Ct-Wert aufweist, deutet dies auf eine
Kontamination des PCR-Laufs hin und er wird als invalide und muss wiederholt werden.
PC :Ct ≤ 38 für S Gen L452R Target (FAMTM)Interne Prozesskontrolle (RNase P):
alle humanen Proben sollten Ct Werte Ct ≤ 38 aufweisen, was auf das
Vorhandensein des humanen RNase P Gens hinweist.

▪
▪

Ein Fehlschlagen des RNase P Nachweises in einer Probe kann ein Hinweis sein auf:
▪
▪
▪
▪
▪

Unsachgemäße Nukleinsäure-Extraktion, was zu Verlust von RNA und/oder RNA Degradation führt.
Fehlen von ausreichend humanem Zellmaterial aufgrund falscher Entnahme oder Verlust der Probenintegrität.
(RT-) PCR-Inhibition.
Unsachgemäßes Reaktions-Setup und Ausführung.
Fehlfunktion von Reagenzien oder Geräten.

Wenn die RNase P Reaktion kein positives Ergebnis für die Proben liefert, muss wie folgt interpretiert werden:▪

Wenn der S Gen-Marker auch bei fehlendem positivem RNase P positiv ist, sollte das Ergebnis als valide angesehen
werden. Es ist möglich, dass einige Proben aufgrund geringer Zellzahlen in der Originalprobe keine RNase PAmplifikation aufweisen. Ein negatives RNase P-Signal schließt das Vorhandensein von SARS-CoV-2 RNA in einer
Probe nicht aus.
Wenn der S Gen Marker UND RNase P für die Probe negativ sind, sollte das Ergebnis für die Probe als invalide bewertet
werden. Wenn eine Restprobe vorhanden ist, sollten das Extraktionsverfahren und der Test wiederholt werden. Wennalle Marker nach dem erneuten Test negativ bleiben, ist das Ergebnis als invalide zu bewerten und es sollte nach
Möglichkeit eine neue Probe entnommen werden.

▪

8.1.2.

ProbenWenn der Testlauf gültig ist, werden die Probenergebnisse wie folgt interpretiert:▪
Positiv (POS):
Ct ≤ 38▪
Negativ (neg):nicht detektiert (ND) oder Ct > 38
Die Sonde zum Nachweis der S-Gen Mutation L452R ist mit FAMTM markiert. Die Sonde für den Nachweis des RNase P Gens
ist mit CY5TM markiert. Die Interpretation der verschiedenen Signalkombinationen ist in der folgenden Tabelle aufgeführt.
Detektionskanal
FAMTM (L452R)

CY5TM (RNAse P)

Positiv

Positiv oder Negativ

Negativ

Positiv

Negativ

Negativ

Interpretation der Ergebnisse
S Gen Mutation L452R detektiert
S Gen Mutation L452R nicht detektiert.
Dieses Ergebnis ist ein Hinweis auf die L452 Wildtyp-Sequenz.
PCR-Inhibition oder Versagen der Reagenzien.
Eine diagnostische Aussage darf nicht getroffen werden.
Der Testlauf sollte wiederholt werden oder es muss eine neue Probeanalysiert werden.

10

9. GRENZEN DES VERFAHRENS
Sehr geringe Mengen an Ausgangsmaterial, z.B. schwach positive Proben (hohe Ct-Werte in der vorangegangenen
diagnostischen SARS-CoV-2 RT-PCR), können zu GSD NovaType Select L452R SARS-CoV-2 Testläufen mit negativen Signalen
führen. Solche Ergebnisse schließen jedoch das Vorhandensein von SARS-CoV-2 RNA nicht aus und somit auch nicht das
Vorliegen einer L452R-Mutation.

10. MARKENZEICHEN UND HAFTUNGSAUSSCHLÜSSE
Die in diesem Dokument verwendeten eingetragenen Namen, Markenzeichen usw. sind als gesetzlich geschützt zu betrachten,
auch wenn sie nicht ausdrücklich als solche gekennzeichnet sind.

11. SICHERHEITSMASSNAHMEN UND WARNHINWEISE
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪
▪

▪

▪

▪

▪
▪

NUR FÜR FORSCHUNGSZWECKE. Nicht zur Verwendung in diagnostischen Verfahren.
Alle Proben humanen Ursprungs sollten als potenziell infektiös betrachtet und behandelt werden.
Reagenzien verschiedener Produktionschargen nicht untereinander austauschen.
Keine Reagenzien anderer Hersteller zusammen mit den Reagenzien dieses Testkits verwenden.
Reagenzien nach dem auf dem Etikett angegebenem Verfallsdatum nicht mehr verwenden.
Beim Umgang mit Proben puderfreie Einweghandschuhe, einen Laborkittel und einen Augenschutz tragen.
Verwenden Sie immer DNase/RNase-freie Einwegreaktionsgefäße und Pipettenspitzen mit Aerosolbarrieren.
Vermeiden Sie eine mikrobielle und Nuklease- (DNase/RNase) Kontamination der Probe und der Kit-Komponenten.
Um eine Kontamination des Arbeitsbereichs mit Nukleinsäuren zu vermeiden, dürfen Reaktionsgefäße/Platten nach der
Amplifikation nicht geöffnet werden.
Die RT-PCR ist hochempfindlich gegenüber Nukleinsäurekontaminationen. Daher muss positives/potenziell positives
Material getrennt von allen anderen Komponenten des Kits gelagert werden.
Belassen Sie Verbrauchsmaterial und Ausstattung in den eindeutig zugewiesenen, getrennten Arbeitsbereichen.
Dieser Assay darf nicht direkt mit der Patientenprobe verwendet werden.
Vor der Verwendung dieses Assays muss die Nukleinsäure mit geeigneten Extraktionsmethoden aus der Originalprobe
extrahiert werden. Beachten Sie hierzu unbedingt die Angaben in der Gebrauchsanleitung des Extraktionskits v.a. in
Bezug auf unterschiedliche Probenmaterialien!
Da Ethanol ein starker Inhibitor der Real-Time PCR ist, muss es vor der Elution der Nukleinsäure während der Extraktion
vollständig eliminiert werden. Bei Verwendung von Spin Columns mit Waschpuffern, die Ethanol enthalten, wird
dringend empfohlen, vor der Elution der RNA einen zusätzlichen Zentrifugationsschritt von 10 min bei ca. 17000 x g(~ 13000 U/min) durchzuführen. Für diesen zusätzlichen Zentrifugationsschritt ist ein neues Sammelröhrchen zu
verwenden.
Das Ergebnis dieses RT-PCR-Kits kann durch potenzielle Mutationen im Genom des Erregers beeinflusst werden, wenndiese in der Primer-/Sondenbindungsregion liegen. Dies kann zu einer Fehleinschätzung führen und/oder den
Erregernachweises erschweren.
RT-PCR-Inhibitoren können eine zu niedrige Bewertung, falsch negative Ergebnisse oder ungültige Testläufe
hervorrufen. Verwenden Sie daher nur Nukleinsäure-Extraktionskits, die RT-PCR-Inhibitoren entfernen und die fürnachgeschaltete RT-PCR-Prozesse bestimmt sind.
Die Real-Time PCR ist nur für qualifiziertes Personal bestimmt, das mit der guten Laborpraxis vertraut und in RT-PCR
geschult ist.
Zur weiteren internen Qualitätskontrolle sollte jedes Labor zusätzlich bekannte Proben verwenden.

11.1. Entsorgungshinweise
Chemikalien und Zubereitungen sind in der Regel Sonderabfälle. Deren Beseitigung unterliegt den nationalen abfallrechtlichen
Gesetzen und Verordnungen. Die zuständige Behörde informiert über die Entsorgung von Sonderabfällen.

12. BESTELLINFORMATIONEN
Produktnummer:

PCOVL452R3PCOVL452R4

GSD NovaType Select L452R SARS-CoV-2
GSD NovaType Select L452R SARS-CoV-2

11(1 x 96 Bestimmungen)(4 x 96 Bestimmungen)

12

13

14

BIBLIOGRAPHY / LITERATUR / BIBLIOGRAPHIE / BIBLIOGRAFIA / BIBLIOGRAFÍA /BIBLIOGRAFIA
European Centre for Disease Prevention and Control. SARS-CoV-2 - increased circulation of variants of concern and vaccine
rollout in the EU/EEA, 14th update – 15 February 2021. ECDC: Stockholm; 2021.
Robert Koch-Institut (2021): RKI - Coronavirus SARS-CoV-2 - SARS-CoV-2: Virologische Basisdaten sowie Virusvarianten.
Available online at https://www.rki.de/DE/Content/InfAZ/N/Neuartiges_Coronavirus/Virologische_Basisdaten.html, updated on
2/23/2021, checked on 3/16/2021.
Volz, Erik; Mishra, Swapnil; Chand, Meera; Barrett, Jeffrey C.; Johnson, Robert; Geidelberg, Lily et al. (2021): Transmission of
SARS-CoV-2 Lineage B.1.1.7 in England: Insights from linking epidemiological and genetic data. In medRxiv,
2020.12.30.20249034. DOI: 10.1101/2020.12.30.20249034.

ABBREVIATIONS / ABKÜRZUNGEN / ABRÉVIATIONS / ABBREVIAZIONI / ABREVIACIÓNES /ABREVIATURAS
Ct/Cq

Threshold cycle (Ct) or quantification cycle (Cq) is the cycle number at which the sample’s reaction curve intersects
the threshold line

NAT

Nucleic acid Amplification Techniques

NTC

No Template Control

15

SYMBOLS KEY / SYMBOLSCHLÜSSEL / EXPLICATION DES SYMBOLES / LEGENDA /SIMBOLOS / TABELA DE SIMBOLOS
Manufactured by / Hergestellt von / Fabriqué par / Prodotto da / Fabricado por / Fabricado por

RUO

FOR RESEARCH USE ONLY. Not for use in diagnostic procedures. / NUR FÜR
FORSCHUNGSZWECKE. Nicht zur Verwendung in diagnostischen Verfahren. / UNIQUEMENTPOUR LA RECHERCHE. Ne pas utiliser dans les procédures de diagnostic. / SOLO DI
RICERCA. Non per l'uso in procedure diagnostiche. / PARA USO EXCLUSIVO ENINVESTIGACIÓN. No debe utilizarse en procedimientos de diagnóstico. / ESCLUSIVAMENTEPARA A INVESTIGAÇÃO. Não para uso em procedimentos de diagnóstico.

LOT

Lot Number / Chargenbezeichnung / Numéro de lot / Lotto / Número de lote / Número de lote
Expiration Date / Verfallsdatum / Date de péremption / Scadenza / Fecha de caducidad / Data de
Validade
Storage Temperature / Lagertemperatur / Température de conservation / Temperatura diconservazione / Temperatura de almacenamiento / Temperatura de Armazenamento
Protect from Light / Vor Licht schützen / Protéger de la Lumière / Proteggere dalla Luce / Proteger
de la Luz / Proteger da Luz

REF

E-MIX

PP│L452R

NFW
PC

Product Number / Produktnummer / Numéro de produit / Numero del prodotto / Número delproducto / Número do produto
Consult Instructions for Use / Arbeitsanleitung beachten / Consulter la notice d’utilisation /Consultare le istruzioni per l’uso / Consulte las Instrucciones de Uso / Consultar as Instruções deUtilização
RT-PCR Enzyme Mix / RT-PCR Enzym-Mix / RT-PCR Mix d'Enzymes / RT-PCR Mix di Enzima /Mescla de Enzimas RT-PCR / Mistura Enzimática RT-PCR
Primer-Probe-Mix(for detection of mutation L452R and for detection of RNase P gene / zum Nachweis der Mutation L452 unddes RNase P Gens / pour la detection de la mutation L452R et pour la détection de la RNase P / per ilrilevamento della mutazione L452R e per il rilevamento del gene RNase P / para la detección de la mutación
L452R y para la detección del gen RNasa P / para detecção da mutação L452R e para detecção do gene
RNase P)

Nuclease free water / Nukleasefreies Wasser / Eau sans nuclease / Acqua priva di nucleosi /Agua libre de nucleasas / Água isenta de Nuclease
Positive Control / Positivkontrolle / Contrôle positif / Controllo positivo / Control positivo / Controle
positivo
Contains sufficient for “n” tests / Ausreichend für “n” Tests / Contenu suffisant pour “n” tests /Contenuto sufficiente per “n” saggi / Contenido suficiente para ”n” tests / Conteúdo suficiente para“n” testes

NovaTec Immundiagnostica GmbH
Waldstraße 23 A6
63128 Dietzenbach, Germany
Tel.:
Email:
Internet:+49 (0) 6074-48760
info@NovaTec-ID.com
www.NovaTec-ID.com

Fax:+49 (0) 6074-487629
PCOVL452Rx_RUO_rev01_20210518

16