BA00295P/00/EN/18.16
71335661

Products

Solutionsvalid from Software version
04.01.zz

Operating Instructions
Cerabar S PMC71, PMP71,
PMP75
Process pressure measurement

ServicesCerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

1.

Order code: XXXXX-XXXXXX
Ser. no.:XXXXXXXXXXXX
Ext. ord. cd.: XXX.XXXX.XX

Serial number

2. www.endress.com/deviceviewer

Endress+Hauser
Operations App

3.

A0023555

Make sure the document is stored in a safe place such that it is always available whenworking on or with the device.
To avoid danger to individuals or the facility, read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures.
The manufacturer reserves the right to modify technical data without prior notice. Your
Endress+Hauser Sales Center will supply you with current information and updates to these
Instructions.

2

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Table of contents
1

Document information . . . . . . . . . . . . . . 4

1.1
1.2
1.3
1.4
1.5

Document function . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . 5
Terms and abbreviations . . . . . . . . . . . . . . . . . . . . . 6
Turn down calculation . . . . . . . . . . . . . . . . . . . . . . . 7

2

7.8
7.9

Scaling the OUT value . . . . . . . . . . . . . . . . . . . . . . 72
System units (SET UNIT TO BUS) . . . . . . . . . . . . . 73

8Maintenance . . . . . . . . . . . . . . . . . . . . . 74

8.1
8.2

Cleaning instructions . . . . . . . . . . . . . . . . . . . . . . . 74
Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . 74

Basic safety instructions . . . . . . . . . . . . . 8

9Troubleshooting . . . . . . . . . . . . . . . . . . 75

2.1
2.2
2.3
2.4
2.5
2.6

Requirements concerning the staff . . . . . . . . . . . . . 8
Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Hazardous area . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

3

Identification . . . . . . . . . . . . . . . . . . . . . 10

3.1
3.2
3.3
3.4

Product identification . . . . . . . . . . . . . . . . . . . . . . 10
Device designation . . . . . . . . . . . . . . . . . . . . . . . . 10
Scope of delivery . . . . . . . . . . . . . . . . . . . . . . . . . . 12
Certificates and approvals . . . . . . . . . . . . . . . . . . 12

9.1
9.2
9.3
9.4
9.5
9.6
9.7
9.8
9.9
9.10

Messages . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Response of outputs to errors . . . . . . . . . . . . . . . . 83
Confirming messages . . . . . . . . . . . . . . . . . . . . . . . 85
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Repair of Ex-certified devices . . . . . . . . . . . . . . . . 86
Spare Parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Software history . . . . . . . . . . . . . . . . . . . . . . . . . . . 87
Hardware history . . . . . . . . . . . . . . . . . . . . . . . . . . 87

10Technical data . . . . . . . . . . . . . . . . . . . . 88

4

Installation . . . . . . . . . . . . . . . . . . . . . . . 13

4.1
4.2
4.3
4.4
4.5

Incoming acceptance, transport, storage . . . . . . 13
Installation conditions . . . . . . . . . . . . . . . . . . . . . 13
General installation instructions . . . . . . . . . . . . . 14
Installation instructions . . . . . . . . . . . . . . . . . . . . 14
Post-installation check . . . . . . . . . . . . . . . . . . . . . 25

5

Wiring . . . . . . . . . . . . . . . . . . . . . . . . . . . 26

5.1
5.2
5.3
5.4

Connecting the device . . . . . . . . . . . . . . . . . . . . . 26
Connecting the measuring unit . . . . . . . . . . . . . . 27
Overvoltage protection (optional) . . . . . . . . . . . . 28
Post-connection check . . . . . . . . . . . . . . . . . . . . . 28

6Operation. . . . . . . . . . . . . . . . . . . . . . . . . 29

6.1
6.2
6.3
6.4
6.5
6.6
6.7
6.8

Onsite display (optional) . . . . . . . . . . . . . . . . . . . 29
Operating elements . . . . . . . . . . . . . . . . . . . . . . . 30
PROFIBUS PA communication protocol . . . . . . . 33
Onsite operation – onsite display connected . . . 53
Endress+Hauser operating program . . . . . . . . . . 56
HistoROM®/M-DAT (optional) . . . . . . . . . . . . . . 56
Locking/unlocking operation . . . . . . . . . . . . . . . 59
Factory setting (reset) . . . . . . . . . . . . . . . . . . . . . 60

7Commissioning . . . . . . . . . . . . . . . . . . . . 62

7.1
7.2
7.3
7.4
7.5
7.6
7.7

Configuring messages . . . . . . . . . . . . . . . . . . . . . 62
Function check . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Commissioning via Class 2 master (FieldCare) . 63
Selecting language and measuring mode . . . . . 63
Position adjustment . . . . . . . . . . . . . . . . . . . . . . . 65
Pressure measurement . . . . . . . . . . . . . . . . . . . . . 66
Level measurement . . . . . . . . . . . . . . . . . . . . . . . . 68

Endress+Hauser

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . 89

3

Document information

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

1Document information

1.1

Document function

These Operating Instructions contain all the information that is required in various phasesof the life cycle of the device: from product identification, incoming acceptance and storage,
to mounting, connection, operation and commissioning through to troubleshooting,
maintenance and disposal.

1.2

Symbols used

1.2.1

Safety symbols

SymbolMeaning

DANGER
A0011189-DE

WARNING
A0011190-DE

CAUTION
A0011191-DE

NOTICE
A0011192-DE

1.2.2

DANGER!
This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inseriousor fatal injury.
WARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inseriousor fatal injury.
CAUTION!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminoror medium injury.
NOTICE!
This symbol contains information on procedures and other facts which do not result inpersonalinjury.

Electrical symbols

SymbolMeaning

Meaning

Direct current

Alternating current

Direct current and alternating current

Ground connection
A grounded terminal which, as far as
the operator is concerned, is grounded
via a grounding system.

Protective ground connection
A terminal which must be connected
to ground prior to establishing anyother connections.

1.2.3

Symbol)Equipotential connection
A connection that has to be connectedto the plant grounding system: Thismay be a potential equalization line ora star grounding system depending on
national or company codes of practice.

Tool symbols

SymbolMeaning
Allen key
A0011221

Hexagon wrench
A0011222

4

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

1.2.4

Document information

Symbols for certain types of information

SymbolMeaning
Permitted
Indicates procedures, processes or actions that are permitted.
A0011182

Forbidden
Indicates procedures, processes or actions that are forbidden.
A0011184

Tip
Indicates additional information.
A0011193

Reference to documentation
A0028658

Reference to pageA
A0028659

Reference to graphic
A0028660

1. , 2. , 3. …

Series of steps

A0031595

Result of a sequence of actions
A0018343

Visual inspection
A0028673

1.2.5

Symbols in graphics

SymbolMeaning

1, 2, 3, 4, ...

1. , 2. , 3. …

Item numbers
Series of steps

A0031595

A, B, C, D, ...

1.2.6

Views

Symbols at the device

SymbolMeaning
Safety instructions
Observe the safety instructions contained in the associated Operating Instructions.
A0019159

1.3

Registered trademarks

KALREZ, VITON, TEFLON
Registered trademarks of E.I. Du Pont de Nemours & Co., Wilmington, USA
TRI-CLAMP
Registered trademark of Ladish & Co., Inc., Kenosha, USA
PROFIBUS PA®
Trademark of the PROFIBUS User Organization, Karlsruhe, Germany
GORE-TEX®
Registered trademarks of W.L. Gore & Associates, Inc., USA

Endress+Hauser

5Document information

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

1.4

Terms and abbreviations

1

2

3

4

p
LRL

0
LRV

URV

URL

MWP

OPL
A0029505

6

Position

Term/Abbreviation

Explanation

1

OPL

The OPL (over pressure limit = sensor overload limit) for the sensors
depends on the lowest-rated element, with regard to pressure, of theselected components, i.e. the process connection must be taken into
consideration in addition to the measuring cell. Also observe pressuretemperature dependency. For the relevant standards and additional notes,
see technical information.
The OPL may be applied for a limited time period.

2

MWP

The MWP (maximum working pressure) for the sensors depends on thelowest-rated element, with regard to pressure, of the selected components,
i.e. the process connection has to be taken into consideration in addition to
the measuring cell. Also observe pressure-temperature dependency. For the
relevant standards and additional notes, see technical information.
The MWP may be applied for an unlimited time.

3

Maximum sensor
measuring range

Range between LRL and URL
This span is the maximum calibratable/adjustable measuring span.

4

Calibrated/Adjusted
measuring span

Range between LRV and URV
Factory setting: 0...URL
Other calibrated spans can be ordered with customised settings.

p-

Pressure-

LRL

Lower range limit-

URL

Upper range limit-

LRV

Lower range value-

URV

Upper range value-

TD

Turn down

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

1.5

Document information

Turn down calculation
1=2

3
URV

LRL LRV

URL
A0029545

Fig. 1:
1Calibrated/Adjusted measuring span
2Zero-based span
3Upper range limit

Example• Calibrated/Adjusted measuring span: 0...5 bar(0...75 psi)
• Lower range value (LRV) = 0 bar• Upper range value (URV) = 5 bar (75 psi)

• Sensor: 10 bar (150 psi)
• Upper range limit (URL) = 10 bar (150 psi)
Turn down (TD):
TD =

TD =

URL
|URV

-

LRV|

10 bar (150 psi)
|5 bar (75 psi)-

0 bar (0 psi)|= 2

In this example, the TD is thus 2:1.
This span is based on the zero point.

Endress+Hauser

7Basic safety instructions

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

2Basic safety instructions

2.1

Requirements concerning the staff

The personnel for installation, commissioning, diagnostics and maintenance must fulfill thefollowing requirements:•Trained, qualified specialists: must have a relevant qualification for this specific function
and task•Are authorized by the plant owner/operator•Are familiar with federal/national regulations•Before beginning work, the specialist staff must have read and understood the instructionsin the Operating Instructions and supplementary documentation as well as in thecertificates (depending on the application)•Following instructions and basic conditions
The operating personnel must fulfill the following requirements:•Being instructed and authorized according to the requirements of the task by the facility'sowner-operator•Following the instructions in these Operating Instructions

2.2

Designated use

The Cerabar S is a pressure transmitter for measuring pressure and level.

2.2.1

Incorrect use

The manufacturer is not liable for damage caused by improper or non-designated use.
Verification for borderline cases:
For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistance inverifying the corrosion resistance of fluid-wetted materials, but does not accept any
warranty or liability.

2.3

Workplace safety

For work on and with the device:•Wear the required personal protective equipment according to federal/national
regulations.
•Switch off the supply voltage before connecting the device.

2.4

Operational safety

Risk of injury!
‣Operate the device in proper technical condition and fail-safe condition only.
‣The operator is responsible for interference-free operation of the device.
Conversions to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers:‣If, despite this, modifications are required, consult with Endress+Hauser.
RepairTo ensure continued operational safety and reliability,‣Carry out repairs on the device only if they are expressly permitted.
‣Observe federal/national regulations pertaining to repair of an electrical device.
‣Use original spare parts and accessories from Endress+Hauser only.

8

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

2.5

Basic safety instructions

Hazardous area

To eliminate a danger for persons or for the facility when the device is used in the hazardous
area (e.g. explosion protection, pressure vessel safety):•Based on the nameplate, check whether the ordered device is permitted for the intended
use in the hazardous area.
•Observe the specifications in the separate supplementary documentation that is an integral
part of these Instructions.

2.6

Product safety

This measuring device is designed in accordance with good engineering practice to meetstate-of-the- art safety requirements, has been tested, and left the factory in a condition inwhich they are safe to operate. It fulfills general safety requirements and legal requirements.
It also conforms to the EC directives listed in the device-specific EC declaration of conformity.
Endress+Hauser confirms this fact by applying the CE mark.

Endress+Hauser

9Identification

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

3

Identification

3.1

Product identification

The following options are available for identification of the measuring device:•Nameplate specifications•Order code with breakdown of the device features on the delivery note
•Enter serial numbers from nameplates in W@M Device Viewer(www.endress.com/deviceviewer): All information about the measuring device isdisplayed.
For an overview of the technical documentation provided, enter the serial number from thenameplates in the W@M Device Viewer (www.endress.com/deviceviewer).

3.2

Device designation

3.2.1

Nameplate•The MWP (maximum working pressure) is specified on the nameplate. This value refers to
a reference temperature of +20 °C (68°F) and may be applied to the device for an unlimited
time. Observe temperature dependency of the MWP. The pressure values permitted at
higher temperatures can be found in the standards EN 1092-1: 2001 Tab. 18 (Withregard to their stability-temperature property, the materials 1.4435 and 1.4404 aregrouped together under 13EO in EN 1092-1 Tab. 18. The chemical composition of the twomaterials can be identical.), ASME B 16.5a – 1998 Tab. 2-2.2 F316, ASME B 16.5a –1998 Tab. 2.3.8 N10276, JIS B 2220.
•The test pressure corresponds to the over pressure limit (OPL) of the device = MWP x 1.5 1).
•The Pressure Equipment Directive (2014/68/EU) uses the abbreviation "PS".
The abbreviation "PS" corresponds to the MWP (maximum working pressure) of themeasuring device.
1)

The equation does not apply for PMP71 and PMP75 with a 40 bar (600 psi) or a 100 bar (1500 psi) measuringcell.

Aluminium and stainless steel housing (T14)123Order code:
Ext. order code:

4Ser. no.:5
6

A0016056

Fig. 2: Nameplate
1Device name
2Order code (for re-orders)
3Extended order code (complete)
4Technical data
5
Serial number (for identification)
6Address of manufacturer

Devices for use in hazardous areas are fitted with an additional nameplate.

10

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Identification

1

2
A0021222

Fig. 3: Additional nameplate
1Approval-specific information
2Document number for safety instructions or drawing number

Devices suitable for oxygen applications or with PVDF process connection are fitted with anadditional nameplate.

1A0022683

Fig. 4: Additional nameplate
1Application limits

Hygenic stainless steel housing (T17)

1
2
3
4
5

Order code:
Ext. ord. cd.:
Ser. no.:7

6A0021552

Fig. 5: Nameplate
1Device name
2Address of manufacturer
3Order code (for re-orders)
4Extended order code (complete)
5
Serial number (for identification)
6Technical data
7Approval-specific information and document number for safety instructions or drawing number

3.2.2

Identifying the sensor type

See parameter "Sensor Meas.Type" in Operating Instructions BA00296P.

Endress+Hauser

11

Identification

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

3.3

Scope of delivery

The scope of delivery comprises:•Cerabar S pressure transmitter•For devices with the "HistoROM/M-DAT" option:
CD-ROM with Endress+Hauser operating program•Optional accessories
Documentation supplied:•The Operating Instructions BA00295P and BA00296P are available via the Internet.
See: www.endress.com Download.
•Brief Operating Instructions KA01022P•Leporello KA00244P•Final inspection report•Also Safety Instructions with ATEX, IECEx and NEPSI devices•Optional: factory calibration form, test certificates

3.4

Certificates and approvals

CE mark, declaration of conformity
The device is designed to meet state-of-the-art safety requirements, has been tested and left
the factory in a condition in which it is safe to operate. The device complies with theapplicable standards and regulations as listed in the EC declaration of conformity and thus
complies with the statutory requirements of the EC Directives. Endress+Hauser confirms the
successful testing of the device by affixing to it the CE mark.

12

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4

Installation

4.1

Incoming acceptance, transport, storage

4.1.1

Incoming acceptance

Installation•Check the packaging and the contents for damage.
•Check the shipment, make sure nothing is missing and that the scope of supply matches
your order.

4.1.2
!

Transport

WARNING

Incorrect transportation
Housing, diaphragm and capillaries may become damaged, and there is a risk of injury!
‣Transport the measuring device to the measuring point in its original packaging or by theprocess connection (with secure transport protection for the diaphragm).
‣Follow the safety instructions and transport conditions for devices weighing more than 18
kg (39.6 lbs).
‣Do not use capillaries as a carrying aid for the diaphragm seals.

4.1.3

Storage

The device must be stored in a dry, clean area and protected against impact (EN 837-2).
Storage temperature range:•–40 °C to +90°C (–40 °F to +194°F)•Onsite display: –40 °C to +85°C (–40 °F to +185°F)•Separate housing: –40 °C to +60°C (–40 °F to +140°F)

4.2

Installation conditions

4.2.1

Dimensions

For dimensions, please refer to "Mechanical construction" section in TI00383P.

Endress+Hauser

13Installation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.3

General installation instructions•Devices with a G 1 1/2 thread:
When screwing the device into the tank, the flat seal has to be positioned on the sealing
surfaceof the process connection. To avoid additional strain on the process isolating diaphragm,
the thread should never be sealed with hemp or similar materials.
•Devices with NPT threads:–Wrap Teflon tape around the thread to seal it.
–Tighten the device at the hexagonal bolt only. Do not turn at the housing.
–Do not overtighten the thread when screwing. Max. torque: 20 to 30 Nm (14.75 to 22.13
lbf ft)•For the following process connections a tightening torque of max. 40 Nm (29.50 lbf ft)
is required:–Thread ISO228 G1/2 (Order option "1A" or "1B")–Thread DIN13 M20 x 1.5 (Order option "1N" or "1P")

4.3.1
!

Mounting sensor modules with PVDF thread

WARNING

Risk of damage to process connection!
Risk of injury!
‣Sensor modules with PVDF thread must be installed with the mounting bracket provided!
!

WARNING

Material fatigue from pressure and temperature!
Risk of injury if parts burst! The thread can become loose if exposed to high pressure andtemperatures.
‣The integrity of the thread must be checked regularly and the thread may need to be retightened with the maximum tightening torque of 7 Nm (5.16 lbf ft). Teflon tape isrecommended for sealing the ½" NPT thread.

4.4

Installation instructions•Due to the orientation of the Cerabar S, there may be a shift in the measured value, i.e.
when the container is empty, the measured value does not display zero. You can correct
this zero point shift either via the "zero" key on the electronic insert, or on the outside ofthe device or via the onsite display. See  ä 30, "Position of operating elements",  ä 31,"Function of operating elements" and  ä 65, "Position adjustment".
•For PMP75, please refer to  ä 17 "Installation instructions for devices with diaphragmseals – PMP75".
•To ensure optimal readability of the onsite display, it is possible to rotate the housing up to380°. See also  ä 24, "Rotating the housing".
•Endress+Hauser offers a mounting bracket for installing on pipes or walls.
See also  ä 21, "Wall and pipe mounting (optional)".

4.4.1

Installation instructions for devices without diaphragm seals –PMP71, PMC71

NOTICE

Damage to the device!
If a heated Cerabar S is cooled during the cleaning process (e.g. by cold water), a vacuum
develops for a short time, whereby moisture can penetrate the sensor through the pressurecompensation (1).

14Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Installation‣If this is the case, mount the Cerabar S with the pressure compensation (1) pointing
downwards.

1
FIELD TERMINALS1

SLANIMRET DLEIF1

SLANIMRET DLEIF•Keep the pressure compensation and GORE-TEX® filter (1) free from contamination andwater.
•Cerabar S devices without a diaphragm seal are mounted as per the norms for a manometer(DIN EN 837-2). We recommend the use of shutoff devices and siphons. The orientation
depends on the measuring application.
•Do not clean or touch process isolating diaphragm seals with hard or pointed objects.
•The device must be installed as follows in order to comply with the cleanability
requirements of the ASME-BPE (Part SD Cleanability):

IF
T DLE
IMRE
SLAN

FIELD

INALS
TERM

SLANIMRET DLEIF

Pressure measurement in gases➀➁

P01-PMx7xxxx-11-xx-xx-xx-001

Fig. 6: Measuring arrangement for pressure measurement in gases
1Cerabar S
2Shutoff device

Mount Cerabar S with shutoff device above the tapping point so that the condensate can flow
into the process.

Endress+Hauser

15

Installation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Pressure measurement in steams

➀
➃

➁

➁

➂
➀
P01-PMx7xxxx-11-xx-xx-xx-002

Fig. 7: Measuring arrangement for pressure measurement in steams
1Cerabar S
2Shutoff device
3U-shaped siphon
4Circular siphon•Use siphons for pressure measurement in steam. The siphon reduces the temperature toalmost ambient temperature. Preferably mount the Cerabar S with the siphon below thetapping point.
Advantages:–defined water column only causes minimal/negligible measured errors–only minimal/negligible thermal effects on the device
Mounting above the tapping point is also possible. Pay attention to the maximum
permitted ambient temperature of the transmitter!
•Fill the siphon with liquid before commissioning.
Pressure measurement in liquids➀➁

P01-PMx7xxxx-11-xx-xx-xx-003

Fig. 8: Measuring arrangement for pressure measurement in liquids
1Cerabar S
2Shutoff device

Mount Cerabar S with shutoff device below or at the same level as the tapping point.

16

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Installation

Level measurement

P01-PMP75xxx-11-xx-xx-xx-000

Fig. 9: Measuring arrangement for level•Mount Cerabar S below the lowest measuring point.
•Do not mount the device at the following positions:
In the fill flow, in the tank outlet or at a point in the container which could be affected bypressure pulses from an agitator.
•Do not mount the device in the suction area of a pump.
•The calibration and functional test can be carried out more easily if you mount the device
after a shutoff device.

4.4.2

Installation instructions for devices with diaphragm seals –PMP75•Cerabar S devices with diaphragm seals are screwed in, flanged or clamped, depending on
the type of diaphragm seal.
•Please note that the hydrostatic pressure of the liquid columns in the capillaries can cause
zero point shift. The zero point shift can be corrected.
•Do not clean or touch the process isolating diaphragm of the diaphragm seal with hard or
pointed objects.
•Do not remove process isolating diaphragm protection until shortly before installation.
NOTICE

Improper handling!
Damage to the device!
‣A diaphragm seal and the pressure transmitter together form a closed, oil-filled calibrated
system. The fill fluid hole is sealed and may not be opened.
‣When using a mounting bracket, sufficient strain relief must be ensured for the capillaries
in order to prevent the capillary bending down (bending radius 100 (3.94 in)).
‣Please observe the application limits of the diaphragm seal filling oil as detailed in theTechnical Information for Cerabar S TI00383P, "Planning instructions for diaphragm seal
systems" section.
NOTICE

In order to obtain more precise measurement results and to avoid a defect in the device,
mount the capillaries as follows:‣Vibration-free (in order to avoid additional pressure fluctuations)‣Not in the vicinity of heating or cooling lines‣Insulate if the ambient temperature is below or above the reference temperature‣With a bending radius of  100 mm (3.94 in).
‣Do not use the capillaries as a carrying aid for the diaphragm seals!

Endress+Hauser

17Installation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Vacuum application
For applications under vacuum, Endress+Hauser recommends mounting the pressuretransmitter below the diaphragm seal. This prevents vacuum loading of the diaphragm sealcaused by the presence of fill fluid in the capillary.
When the pressure transmitter is mounted above the diaphragm seal, the maximum height
difference H1 in accordance with the illustrations below must not be exceeded.

FIELD TERMINALS

H1

A0020472

Fig. 10: Installation above the lower diaphragm seal

The maximum height difference depends on the density of the filling oil and the smallest
ever pressure that is permitted to occur at the diaphragm seal (empty vessel), see illustration
below:
[m]
12.0
1
10.0
2
8.0
3
A 6.0
4
4.0
5

2.0
0.0
50 100 200

300

400

500

600

700

800

900 1000[mbarabs]

B
A0023986-en

Fig. 11: Diagram of maximum installation height above the lower diaphragm seal for vacuum applications depending on the pressure
at the diaphragm seal
AHeight difference H1
BPressure at diaphragm seal
1Low temperature oil
2Vegetable oil
3Silicone oil
4High-temperature oil
5Inert oil

18

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Installation

Mounting with temperature isolator
Endress+Hauser recommends the use of temperature isolators in the event of constantextreme medium temperatures which lead to the maximum permissible electronics
temperature of +85 °C (+185°F) being exceeded. Depending on the filling oil used,diaphragm seal systems with temperature isolators can be used for maximum temperaturesof up to +400 °C (+752 °F). For the temperature application limits, see technical
Information, "Diaphragm seal filling oils" section. To minimize the influence of rising heat,
Endress+Hauser recommends the device be mounted horizontally or with the housing
pointing downwards. The additional installation height also brings about a maximum zero
point shift of 21 mbar (0.315 psi) due to the hydrostatic column in the temperature isolator.
You can correct this zero point shift at the device.
A

B
1

Tp

C
1

Tp

Ta

1
Tp

Ta

Ta

2

2

[°F]

Tp

[°C]

A
+752

+400

+662

+350

+572

+300

+482

+250

+392

+200

+302

+150

+212

+100

+122

+50

+32

0

-58

-50

-148

-100
-60
-76

-40
-40

-20
-4

0
+32

+20
+68

+40
+104

C

+60

B

+80

+140

+176

+100
+212

[°C]
[°F]

Ta
A0032475

Fig. 12:1Transmitter
2Insulation material

Position

Insulation

Ambient temperature (Ta) at transmitter

Maximum process temperature (Tp)A

No insulation

60 °C (140 °F)

400 °C (752 °F) 1)

85 °C (185 °F)

200 °C (392 °F)

80 °C (176 °F)

400 °C (752 °F) )

85 °C (185 °F)

300 °C (572 °F)

67 °C (153 °F)

400 °C (752 °F) )

85 °C (185 °F)

200 °C (392 °F)

BC

30 mm (1.18 in) Insulation

Maximum insulation

1) Process temperature: max. 400 °C (752 °F), depending on the diaphragm seal filling oil used

Endress+Hauser

19Installation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.4.3

Seal for flange mounting

NOTICE

Corrupted measurement results.
The seal is not allowed to press against the process isolating diaphragm as this could affectthe measurement result.
‣Ensure that the seal is not touching the process isolating diaphragm.

1

2

A0017743

Fig. 13:
1Process isolating diaphragm
2Seal

4.4.4

Installation with heat insulation – PMC71 high temperatureversion and PMP75

The devices must only be insulated up to a certain height.
The maximum permitted insulation height is labeled on the devices and applies to an
insulation material with a specific heat conductivity and to the maximum permitted ambientand process temperature (see the following table). The data were determined under themost critical application "quiescent air".

Insulationmaterial

TA

TP
P01-PMx7xxxx-11-xx-xx-xx-010

Fig. 14: Maximum permitted insulation height, here e.g. PMC71 with flange

PMC71

PMP75

Ambient temperature (TA) 70 °C (158°F)

70 °C (158°F)

Process temperature (TP)150 °C (302°F)

 400 °C (752°F)1)

Heat conductivity
insulation material

 0.04 W/(m x K)

1) Depending on the diaphragm seal filling oil used (see Technical Information TI00383P Cerabar S)

20

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.4.5

Installation

Wall and pipe mounting (optional)

Endress+Hauser offers a mounting bracket for installation on pipes or walls (for pipe
diameters from 1 ¼" to 2").

122 (4.8)
158 (6.22)

ø6 (0.24)

52 (2.05)

86 (3.39)

70 (2.76)

ø42...60 (1.65...2.36)

140 (5.51)
175 (6.89)
A0028493

Engineering unit mm (in)Please note the following when mounting:•Devices with capillary tubes: mount capillaries with a bending radius 100 mm (3.94 in).
•When mounting on a pipe, tighten the nuts on the bracket uniformly with a torque of atleast 5 Nm (3.69 lbs ft).

Endress+Hauser

21

Installation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.4.6

Assembling and mounting the "separate housing" version

6
5
4

7

3
2

1

r ³ 120 (4.72)

mm (in)P01-PMx7xxxx-11-xx-xx-xx-011

Fig. 15: "Separate housing" version
1In the "separate housing" version, the sensor is supplied with process connection and cable fitted.
2Cable with connection jack
3Pressure compensation
4Plug
5Locking screw
6Housing fitted with housing adapter, included
7Mounting bracket suitable for wall and pipe mounting, included

Assembly and mounting
1.Connect plug (item 4) into the corresponding connection jack of the cable (item 2).
2.Plug the cable into the housing adapter (item 6).
3.Tighten the locking screw (item 5).
4.Mount the housing on a wall or pipe using the mounting bracket (item 7).
When mounting on a pipe, tighten the nuts on the bracket uniformly with a torque ofat least 5 Nm (3.69 lbf ft).
Mount the cable with a bending radius (r) 120 mm (4.72 in).

22

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.4.7

Installation

PMP71, version prepared for diaphragm seal mount –welding recommendation

1

2

3

ø2.5 (0.1)ø7.95 (0.31)

5 (0.2)

A1

A0028495

Fig. 16:
1
2
3
A1

Version U1: prepared for diaphragm seal mount
Setscrew
Bearing
Hole for fill fluid
See the "Welding recommendation" table below

Engineering unit mm (in)Endress+Hauser recommends welding on the diaphragm seal as follows for the "XSJ Prepared for diaphragm seal mount" version in feature 110 "Process connections" in the ordercode up to, and including, 40 bar (600 psi) sensors: the total welding depth of the fillet weld
is 1 mm (0.04 in) with an outer diameter of 16 mm (0.63 in). Welding is performed
according to the WIG method.
Consecutive seamno.
A1
for sensors
 40 bar (600 psi)

Sketch/welding groove shape,
dimension as per DIN 8551

s1 a0.8

Base material matching

Welding method
DIN EN ISO 24063

Welding
position

Inert gas,
additives

Adapter made of AISI 316L
(1.4435) to be welded todiaphragm seal made of
AISI 316L (1.4435 or 1.4404)

141

PB

Inert gas
Ar/H 95/5
Additive:
ER 316L Si(1.4430)

A0024811

Information on filling
The diaphragm seal must be filled as soon as it has been welded on.
•After welded into the process connection, the sensor assembly must be properly filled witha filling oil and sealed gas-tight with a sealing ball and lock screw.
Once the diaphragm seal has been filled, at the zero point the device display should notexceed 10% of the full scale value of the cell measuring range. The internal pressure of thediaphragm seal must be corrected accordingly.
•Adjustment / calibration:–The device is operational once it has been fully assembled.
–Perform a reset. The device must then be calibrated to the process measuring range as
described in the Operating Instructions.

Endress+Hauser

23

Installation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.4.8

Rotating the housing

The housing can be rotated up to 380° by loosening the Allen screw.
T14

T15

T17

2

3
A0019996

1.T14 housing: Loosen setscrew with a 2 mm (0.08 in) Allen key.
T15 andT17 housing: Loosen setscrew with a 3 mm (0.12 in) Allen key.
2.Rotate housing (max. up to 380 °).
3.Retighten setscrew with 1 Nm (0,74 lbf ft.

4.4.9

Closing the housing cover

NOTICE

Devices with EPDM cover seal - transmitter leakiness!
Mineral-based, animal-based or vegetable-based lubricants cause the EPDM cover seal toswell and the transmitter to become leaky.
‣The thread is coated at the factory and therefore does not require any lubrication.
NOTICEThe housing cover can no longer be closed.
Damaged thread!
‣When closing the housing cover, please ensure that the thread of the cover and housing
are free from dirt, e.g. sand.If you feel any resistance when closing the cover, check thethread on both again to ensure that they are free from dirt.
Close cover on a hygenic stainless steel housing (T17)

P01-PMD75xxx-17-xx-xx-xx-000

Fig. 17: Close cover

The covers for the terminal and electronics compartment are hooked into the casing andclosed with a screw. These screws should be finger-tightened (2 Nm (1.48 lbf ft)) to the stop
to ensure that the covers sit tightly.

24

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

4.5

Installation

Post-installation check

After installing the device, carry out the following checks:•Are all screws firmly tightened?
•Are the housing covers screwed down tight?

Endress+Hauser

25

Wiring

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

5

Wiring

5.1

Connecting the device!

WARNING

Risk of electric shock!
If the operating voltage is > 35 VDC: Dangerous contact voltage at terminals.
‣In a wet environment, do not open the cover if voltage is present.
!

WARNING

Limitation of electrical safety due to incorrect connection!
•Risk of electric shock and/or explosion in hazardous areas! In a wet environment, do notopen the cover if voltage is present.
•When using the measuring device in hazardous areas, installation must comply with thecorresponding national standards and regulations and the Safety Instructions or
Installation or Control Drawings.
•Devices with integrated overvoltage protection must be earthed.
•Protective circuits against reverse polarity, HF influences and overvoltage peaks are
installed.
•The supply voltage must match the supply voltage on the nameplate. (See also  ä 10,"Nameplate".)•Switch off the supply voltage before connecting the device.
•Remove housing cover of the terminal compartment.
•Guide cable through the gland. For cable specification see  ä 28, "Cable specification".
•Connect device in accordance with the following diagram.
•Screw down housing cover.
•Switch on supply voltage.

➀

➃

PA

PA

➁

➄
PA

PA

➂
P01-xMx7xxxx-04-xx-xx-xx-001

Fig. 18: Electrical connection of PROFIBUS PA
Please refer also to  ä 27, "Supply voltage".
1Housing
2Internal earth terminal
3External earth terminal
4Supply voltage, for version in non-hazardous area = 9 to 32 V DC
5Devices with integrated overvoltage protection are labeled OVP (overvoltage protection) here.

26

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

5.1.1

WiringConnecting devices with an M12 connector

PIN assignment for M12 connector

4

3

1

2

PIN

Meaning

1

Signal +2

Not assigned

3Signal –4

Earth

A0011175

5.1.2

Connecting devices with 7/8" plug

PIN assignment for 7/8" connector

1

3

2

4

PIN

Meaning

PIN

Meaning

1

Signal –2

Signal +3

Shield4

Not assigned

A0011176

Connecting the cable version

rd
bk
gnye

+
–

+

–

5.1.3

PE
4...20 mA
P01-PMx4xxxx-04-xx-xx-xx-010

Fig. 19: rd = red, bk = black, gnye = green-yellow

5.2

Connecting the measuring unit

For further information on the network structure and grounding and for further bus systemcomponents such as bus cables, see the relevant documentation, e.g. Operating Instructions
BA00034S "Guidelines for planning and commissioning PROFIBUS DP/PA" and the
PNO Guideline.

5.2.1

Supply voltage

Version for non-hazardous area: 9 to 32 V DC!

WARNING

Supply voltage might be connected!
Risk of electric shock and/or explosion!
‣When using the measuring device in hazardous areas, installation must comply with thecorresponding national standards and regulations and the Safety Instructions or
Installation or Control Drawings.
‣All explosion protection data are given in separate documentation which is available uponrequest. The Ex documentation is supplied as standard with all devices approved for use
in explosion hazardous areas.

Endress+Hauser

27

Wiring

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

5.2.2

Current consumption

Up to HW Version 1.10: 11 mA ±1 mA, switch-on current corresponds to IEC 61158-2,
Clause 21.
As of HW Version 02.00: 13 mA ±1 mA, switch-on current corresponds to IEC 61158-2,
Clause 21.
As of Hardware Version 1.10, you will find a label in the device on the electronic insert.

5.2.3

Terminals•Supply voltage and internal ground terminal: 0.5 to 2.5 mm2 (20 to 14 AWG)
•External ground terminal: 0.5 to 4 mm2 (20 to 12 AWG)

5.2.4

Cable specification•Use a twisted, shielded two-wire cable, preferably cable type A.
•Outer cable diameter: 5 to 9 mm (0.2 to 0.35 in)
For further information on the cable specifications, see Operating Instructions BA00034S"Guidelines for planning and commissioning PROFIBUS DP/PA", PNO Guideline 2.092"PROFIBUS PA User and Installation Guideline" and IEC 61158-2 (MBP).

5.2.5

Grounding and shielding

Cerabar S must be earthed, for example by means of the external earth terminal.
Different grounding and shielding installation methods are available for PROFIBUS PA
networks such as:•Isolated installation (see also IEC 61158-2)•Installation with multiple grounding•Capacitive installation

5.3

Overvoltage protection (optional)

NOTICE

Device could be destroyed!
Devices with integrated overvoltage protection must be earthed.
Devices showing version "M" in feature 100 "Additional options 1" or feature 110 "Additional
options 2" in the order code are equipped with overvoltage protection (see also Technical
Information TI00383P "Ordering information".
•Overvoltage protection:–Nominal functioning DC voltage: 600 V–Nominal discharge current: 10 kA•Surge current check î = 20 kA as per DIN EN 60079-14: 8/20 s satisfied•Arrester AC current check I = 10 A satisfied

5.4

Post-connection check

Perform the following checks after completing electrical installation of the device:•Does the supply voltage match the specifications on the nameplate?
•Is the device properly connected?
•Are all screws firmly tightened?
•Are the housing covers screwed down tight?
As soon as voltage is applied to the device, the green LED on the electronic insert lights up
for a few seconds or the connected onsite display lights up.

28

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6

Operation

Operation

Feature 20 "Output; operation" in the order code provides you with information on theoperating options available to you.
Versions in the order code

Operation

M

PROFIBUS PA; external and LCD

Via onsite display and 1 key on the exterior of thedeviceN

PROFIBUS PA; internal and LCD

Via onsite display and 1 key on the inside of the device

O

PROFIBUS PA; internal

Without onsite display, 1 key on the inside of the device

6.1

Onsite display (optional)

A 4-line liquid crystal display (LCD) is used for display and operation. The onsite display
shows measured values, fault messages and notice messages. The display of the device canbe turned in 90° steps. Depending on the installation position of the device, this makes it
easy to operate the device and read the measured values.
Functions:•8-digit measured value display including sign and decimal point, unit display.
•Bar Graph as graphic display of standardized value of the Analog Input Block(see also graphic,  ä 72, "Scaling the OUT value").
•Simple and complete menu guidance thanks to separation of the parameters into several
levels and groups.
•Menu guidance in 8 languages (de, en, fr, es, it, nl, jp, ch)•Each parameter is given a 3-digit ID number for easy navigation.
•Option for configuring the display according to individual requirements and desires, suchas language, alternating display, contrast setting, display of other measured values suchas sensor temperature.
•Comprehensive diagnostic functions (fault and warning message).
•Rapid and safe commissioning with the Quick Setup menus.
Measured value display
Function nameValue

Parameter
Identification
numberHeader line
Main line
Information
line
–

+

E

Unit
SymbolBargraph

Editing modes
Operating keys
Selection
options

Value thatcan be edited

Current measured value
P01-xxxxxxxx-07-xx-xx-xx-011

Endress+Hauser

29

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

The following table illustrates the symbols that can appear on the onsite display. Foursymbols can occur at one time.
SymbolMeaning
Alarm symbol– Symbol flashing: warning, device continues measuring.
– Symbol permanently lit: error, device does not continue measuring.
Note: The alarm symbol may overlie the tendency symbol.
Lock symbol
The operation of the device is locked.
Unlock device, see  ä 59, "Locking/unlocking operation".
Communication symbol
Data transfer via communication.
Tendency symbol (increasing)
The primary value of the Transducer Block is increasing.
Tendency symbol (decreasing)
The primary value of the Transducer Block is decreasing.
Tendency symbol (constant)
The primary value of the Transducer Block has remained constant over the past few
minutes.

6.2

Operating elements

6.2.1

Position of operating elements

With regard to aluminum and stainless steel housings (T14), the operating key is locatedeither outside the device under the protection cap or inside on the electronic insert. In
hygienic stainless housings (T17), the operating key is always located inside on theelectronic insert. Additionally, three operating keys are located on the optional onsite
display.

➅➆
on

t

➀

➀
off

➂

➁

➃

0%

➄

Zero
Display

0%
Zero

2

on

on

ON

off

off

1

2

3

4

CK S D A 0

8

5

8

6

7

1 2 3 4 5 6 7 8
AddressSW
HW

P01-PMx7xxxx-19-xx-xx-xx-075

Fig. 20:

Operating key external, under the protective flap

1Operating key for position adjustment (zero point correction)and total reset

30

PC

1

Histo ROM

Sensor

P01-xxxxxxxx-19-xx-xx-xx-105

Fig. 21:

Operating key and operating elements, internal

1Green LED to indicate value is accepted
2Operating key for position adjustment (zero point correction)and total reset
3DIP switch for hardware address
4Slot for optional display
5Slot for optional HistoROM®/M-DAT
6DIP-switch for locking/unlocking measured-value-relevant
parameters
7DIP-switch for damping on/off

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.2.2

Operation

Function of operating elements

Operating elements0%Zero
P01-xxxxxxxx-19-xx-xx-xx-107

Meaning– Position adjustment (zero point correction): Press key for at least 3 seconds. If the
LED on the electronic insert lights up briefly, the pressure applied has been
accepted for position adjustment.
See also the following section "Performing position adjustment on site".
– Total reset: Press key for at least 12 seconds. If the LED on the electronic insert
lights up briefly, the reset is being carried out.
Set address in the bus. See also  ä 36, "Device identification and addressing".

on
off

ON

01

1

2

3

4

CK S D A 0

8

5

8

6

7

1 2 3 4 5 67 8

AddressSW
HW

P01-xxxxxxxx-19-xx-xx-xx-109

1

τ

on

2

off

P01-xxxxxxxx-19-xx-xx-xx-108– DIP-switch 1: for locking/unlocking measured-value-relevant parameters
Factory setting: off (unlocked)
See also  ä 59, "Locking/unlocking operation".
– DIP switch 2: damping on/off
Factory setting: on (damping on)

Performing position adjustment on site
•The operation must be unlocked. See  ä 59, "Locking/unlocking operation".
•The device is configured for the Pressure measuring mode as standard. You can switch
measuring modes by means of the MEASURING MODE parameter. See  ä 63, "Selecting
language and measuring mode".
•The pressure applied must be within the nominal pressure limits of the sensor. See
information on the nameplate.
Carry out position adjustment:
1.Pressure is present at device.
2.Press key for at least 3 seconds.
3.If the LED on the electronic insert lights up briefly, the pressure applied has been acceptedfor position adjustment.
If the LED does not light up, the pressure applied was not accepted. Observe the input
limits. For error messages see  ä 75, "Messages".

Endress+Hauser

31Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.2.3

Function of the operating elements – onsite display connected

Operating key(s)O
SF

Meaning– Navigate upwards in the picklist– Edit the numerical values and characters within a function– Navigate downwards in the picklist– Edit the numerical values and characters within a function– Confirm entry– Jump to the next item

O and F
S and F

O and S

Contrast setting of onsite display: darker

Contrast setting of onsite display: brighter

ESC functions:– Exit edit mode without saving the changed value.
– You are in a menu within a function group. The first time you press the keys
simultaneously, you go back a parameter within the function group. Each time youpress the keys simultaneously after that, you go up a level in the menu.
– You are in a menu at a selection level. Each time you press the keys
simultaneously, you go up a level in the menu.
Note:
The terms function group, level and selection level are explained  ä 53, "General
structure of the operating menu".
Set address in the bus. See also  ä 36, "Device identification and addressing".

on
off

ON

01

1

2

3

4

CK S D A 0

8

5

8

6

7

1 2 3 4 5 67 8

AddressSW
HW

P01-xxxxxxxx-19-xx-xx-xx-109

32

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

6.3

PROFIBUS PA communication protocol

6.3.1

System architecture➀➁➂
PROFIBUS DP
ENDRESS + HAUSER➄PROFIBUS PA

Deltabar S

Deltapilot S

T

Cerabar S➃

P01-xxxxxxxx-14-xx-xx-xx-001

Fig. 22: PROFIBUS system architecture
1PC with PROFIBUS interface card (Profiboard/Proficard) and operating program FieldCare (Class 2 master)
2PLC (Class 1 master)3Segment coupler (DP/PA signal converter and bus feed unit)
4Other measuring devices and adjusters such as valves
5PROFIBUS PA terminating resistor

Further information on PROFIBUS PA can be found in Operating Instructions BA00034S"Guidelines for planning and commissioning PROFIBUS DP/PA", the PNO Guideline andstandards IEC 61158, IEC 61784, EN 50170/DIN 19245 and EN 50020 (FISCO model).

6.3.2

Number of devices•Endress+Hauser Cerabar S devices meet the requirements of the FISCO model.
•Due to the low current consumption, the following can be operated at one bus segment
when installation is performed according to FISCO:
Up to HW Version 1.10:–Up to 9 Cerabar S for Ex ia, CSA and FM IS applications–Up to 32 Cerabar S in all other applications, e.g. in non-Ex areas, Ex nA etc.
As of HW Version 02.00:–Up to 7 Cerabar S for Ex ia, CSA and FM IS applications–Up to 27 Cerabar S in all other applications, e.g. in non-Ex areas, Ex nA etc.
The maximum number of measuring devices at one bus segment is defined by their current
consumption, the performance of the bus coupler, and the required bus length.
As of Hardware Version 1.10, you will find a label in the device on the electronic insert.

Endress+Hauser

33

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.3

Operation

Special configuration and operating programs from various manufacturers are available for
configuring the device, such as the Endress+Hauser operating program FieldCare (see ä 56, "Endress+Hauser operating program"). This operating program makes it possible toconfigure PROFIBUS PA and the device-specific parameters. The predefined function blocks
allow uniform access to all the network and device data.

6.3.4

Identification number of the device

The "IDENT NUMBER SEL." parameter allows users to modify the identification number.
The identification number "IDENT NUMBER SEL." must support the following settings:
Values for "IDENT NUMBER
SEL."

Description

0"0x9700"

Profile-specific transmitter identification number with the "Classic" or"Condensed" status.

1"0x1541"

Identification number for the new Cerabar S generation (PMC71, PMP71,
PMP75).

127"Auto. Id. Num."

Device adaptation mode (the device can communicate using a variety ofidentification numbers), see "Smart device management" (automatic smart
device management).

128"0x1501"

Compatibility mode for the old Cerabar S generation (PMC731, PMP731,
PMC631, PMP635).

The "Automatic Identification Number Selection" (value = 127) for Profile 3.02 is describedin the section on smart device management (automatic smart device management).
The choice of identification number affects the status and diagnostic messages ("Classic" or"Condensed"). "Old" identification numbers work with the "Classic" status and old diagnostic
messages.
Depending on the configuration data of the user or the behavior selected in the
COND.STATUS DIAG "Physical" block parameter, new identification numbers and the profile
identification number work with either the "Condensed" or "Classic" status.
The identification number can only be changed if no cyclic communication is taking placewith the device.
Cyclic data transmission and the corresponding identification number of the device remain
the same until cyclic transmission is aborted and reestablished or the device is shut down.
When reestablishing cyclic data transmission the device uses the last identification number.
The choice of identification number also determines how many modules are assigned during
cyclic communication. All blocks are internally instantiated in advance for all the devices but
only the configured modules can be accessed depending on the entries in the device master
data.
Table of the function blocks:

34"IDENT NUMBER
SEL." parameter0128(Profile-specific) (Oldidentification
number)127(Auto.
identification
number)

1(New identification
number)Cerabar S

3 blocks(PB,TB,AI)...

3 Blocks (PB, TB, AI)

1 module(1xAI)...

Depends on theidentification
numberautomaticallyselected.

1 module(1x AI)

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Table of the identification numbers:
Value for "IDENTNUMBER SEL."

Identification
number

Selection text

StatusDiagnosis

0(Profile-specific 3.x)0x97000x9700Classic status /New diagnostic messages
Condensed status

128(Old identification
number)0x15010x1501

Classic status

Old diagnostic messages

127(Adaptation mode)

0x9700/0x1501/
0x1541

Auto.
Identification
number"Auto Id. Num."

Depends on ID
numbers

Depends on ID numbers

1(New identification
number)0x15410x1541

Classic status /New diagnostic messages
Condensed status

Smart device management (automatic smart device management)
Smart PA device management is performed by automatically adapting the device
identification number. This makes it possible to replace old devices with new models without
having to modify the PLC, allowing the transition from an installed device technology to amore sophisticated technology without interrupting the process.
With the "Automatic Identification Number Selection" option, the device behavior and rules(diagnostics, cyclic communication etc.) remain the same as those for a static identification
number. The identification number is selected automatically depending on the recognizedrequest frame -"Set Slave Parameter" or "Set Slave Address".
It is permitted to change the identification number in two specific device transition states,
namely in the adaptation mode and only if the identification number is listed in the table
above.
If the identification number is undefined and the selector is set to "Auto Id. Num.", followinga "Get Slave Diagnose" frame the device returns an identification number diagnostic value
which is compatible with the device. After every new "Get Slave Diagnose" frame, the device
returns another identification number that is compatible with the device until the PLC sends
a "Set Slave Address" or "Set Slave Parameter" frame with a known identification number.

Endress+Hauser

35

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.5

Device identification and addressing

Note the following points:•An address must be assigned to every PROFIBUS PA device. Only when the address isconfigured correctly will the device be recognized by the control system/master.
•Each address may only be assigned once in each PROFIBUS PA network.
•Valid device addresses are in the range from 0 to 125.
•The address 126 set at the factory can be used to check the function of the device and toconnect to a PROFIBUS PA network already in operation. This address then has to bechanged in order to integrate additional devices.
•All devices have the address 126 and software addressing on leaving the factory.
•The FieldCare operating program is delivered with the address 0 (default setting).
There are two ways of assigning the device address to Cerabar S:•Using a DP Class 2 master operating program, such as FieldCare or•On site using the DIP switches

➀

➁

0%
Zero
Display

2

ON

off

off

1

2

3

4

CK S D A 0

8

5

8

6

7

2 + 8 = 10
PC

1

on

1 2 3 4 5 6 7 8
Address–+

1
2

250002272-–

SW
HW

HW-Version:
SW-Version:

Histo ROM

Sensor

on

ON

on

E

off

01

1

2

3

4

CK S D A

08

5

8

6

7

1 2 3 4 5 6 7 8
AddressSW
HW
P01-xxxxxxxx-19-xx-xx-xx-112

Fig. 23: Configuring the device address using the DIP switches
1If necessary, remove onsite display (optional)
2Set the hardware address via the DIP switches

Hardware addressing
Hardware addressing is configured as follows:
1.Set DIP switch 8 (SW/HW) to "Off".
2.Configure the address with DIP switches 1 to 7 (see graphic above).
3.You have to wait 10 seconds for a change in address to take effect. The device is restarted.
DIP-switch

1

2

3

4

5

6

7

Weighting in "On" position

1

2

4

8

16

32

64

Weighting in "Off" position

0

0

0

0

0

0

0

Software addressing
Software addressing is configured as follows:
1.Set DIP switch 8 (SW/HW) to "On" (factory setting).
2.The device is restarted.
3.The device reports its current address. Factory setting: 126
4.Configure the address via the configuration program.
See the following section for entering a new address using the FieldCare.
For other operating programs, please refer to the relevant Operating Instructions.

36

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Configuring the new address using the FieldCare. DIP switch 8 (SW/HW) is set to "On" (SW):
1.Using the "Device Operation" menu  select the "Connect" option. The "Open Connection
Wizard" screen is displayed.
2.The device reports its current address. Factory setting: 126 1))
3.The device has to be disconnected from the bus before you can assign the device a new
address. For this purpose, select the "Disconnect" option in the  "Device Operation"
menu.
4.Using the "Device Operation" menu, select  "Device Functions"  "Additional Functions" "Set Device Station Address". The "PROFIdtm DPV1 (Set Device Station address)"screen is displayed.
5.Enter the new address and confirm with "Set".
6.The new address is assigned to the device.

1)

The address 126 is not adjustable via the menu. After a reset (code 2712) the address is stored as a default address in the device.

Endress+Hauser

37

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.6

System integration

Device Master Files (GSD)
The device is ready for system integration once commissioning has been effected via theClass 2 master (FieldCare). To integrate the field devices into the bus system, the PROFIBUS
PA system requires a description of the device such as device identification, ID number,supported communication features, module structure (combination of cyclic input andoutput telegrams) and meaning of diagnosis bits.
These data are contained in a Device Master File (GSD file) which is placed at the disposal of
the PROFIBUS DP master (e.g. PLC) while the communication system is being commissioned.
Device bitmaps, which appear as icons in the network tree, can also be integrated.
The following versions of GSD are possible when using devices that support the "PA devices"
profile:•Manufacturer-specific GSD, ID number: 0x1541:
This GSD guarantees the unlimited functionality of the field device.
All device-specific process parameters and functions are available.
•Manufacturer-specific GSD, ID number: 0x1501:
The device behaves like a Cerabar S PMC731, PMP731, PMC631, PMP635.
See Operating Instructions BA00168P.
•Profile GSD:
As an alternative to the manufacturer-specific GSD, the PNO makes a general database fileavailable with the name PA139700.gsd for devices with an Analog Input Block. This file
supports the transmission of the primary value. The transmission of a 2ND CYCLIC VALUE
or a display value is not supported. If a system has been configured with profile GSDs, it ispossible to exchange devices that are supplied by various manufacturers.
The following Device Master Files (GSD) can be used with Cerabar S:
Name of device

Comments

ID number
GSD(IDENT_NUMBER_SELECT)1)

Cerabar S
PROFIBUS PA

Profile GSD

0x9700Device-specific GSD

2)Type fileBit map

PA139700.gsd

0x1541

EH3x1541.gsd
EH021541.gsd3)

0x15012
Device-specific GSD,
the device behaves likea Cerabar S PMC731,
PMP731, PMC631,
PMP635.
See Operating
Instructions
BA00168P.

EH3_1501.gsd
EH3x1501.gsd

EH_1541_d.bmp/.dib
EH_1541_n.bmp/.dib
EH_1541_s.bmp/.dip
EH31501x.200

EH_1501_d.bmp/.dib
EH_1501_n.bmp/.dib
EH_1501_s.bmp/.dip

1) Select the corresponding ID number by means of the IDENT_NUMBER_SEL parameter
menu path FieldCare: PROFILE VIEW PHYSICAL BLOCK PB PARAMETER
menu path onsite display: GROUP SELECTION OPERATING MENU TRANSMITTER INFO PA DATA.
2) Each device receives an ID number from the Profibus User Organization (PNO). The name of the Device Master File (GSD) is derived from this.
For Endress+Hauser, this ID No. starts with the manufacturer ID "15xx".
3) The Profile 3.02 GSD file with the "Condensed Status" option is only compatible with SW 04.01.zz and must be imported individually into the configuration tool.

It is only possible to change the "IDENT NUMBER SEL" parameter if either the device is not
integrated into the cyclic communication (not planned in the PLC) or the cyclic
communication of the PLC is at Stop. If an attempt is nevertheless made to change theparameter via configuration software, e.g. FieldCare, the input is ignored.

38

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

The Device Master Files (GSD) for Endress+Hauser devices can be acquired in the following
manner:•Internet Endress+Hauser: http://www.de.endress.com Download Search for "GSD"•Internet PNO: http://www.profibus.com (Products – Product Guide)•On CD-ROM from Endress+Hauser, order number: 56003894
The Profile Device Master Files (GSD) of the PNO can be acquired in the following manner:•Internet PNO: http://www.profibus.com (Products – Profile GSD Library)
Directory structure of GSD files from Endress+Hauser
For Endress+Hauser field devices with PROFIBUS PA interface, all the data which are needed
for configuration are contained in a compressed file. After unpacking the file, the following
structure is generated:
Cerabar_S/PA/Profile3/Revision1.0/
BMP/
Eh1541_d.bmp
Eh1541_n.bmp
Eh1541_s.bmp
DIB/
Eh1541_d.dib
Eh1541_n.dib
Eh1541_s.dib
GSD/
Eh3x1541.gsd/EH021541.gsd
Info/
Liesmich.pdf
Readme.pdf
P01-PMx7xxxx-02-xx-xx-xx-003

Fig. 24: Directory structure of GSD 1541

•Revision x.x stands for the corresponding device version.
•Information relating to the implementation of the field transmitter and any dependencies
in the device software can be found in the "Info" folder. Read this information carefully
before configuring.
•Device-specific bitmaps can be found in the directories "BMP" and "DIB". The utilization ofthese will depend on the configuration software that is being used.
Working with Device Master Files (GSD)
The Device Master Files (GSD) must be integrated into a specific subdirectory of the
PROFIBUS DP configuration software of the PLC used. Depending on the software that isbeing used, these files can be copied to the program-specific directory or can be read into the
database using the import function within the configuration software.
Detailed information on the directories to which the Device Master Files (GSD) are to be
saved is provided in the description of the configuration software used.

Endress+Hauser

39

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.7

Cyclic data exchange

Cerabar S block model

Measured variable

SensorSignal evaluation

Display
with scaling

Physical Block
Device-specific characteristics,
e.g. software revision

SEL. DISPLAY VAL.
PA Input value
Main Process
Value

Primary Value (PV)MEASURED
VALUE
Transducer Block
Adjusting the measured value

Analog Input Block
Parameter for processing
the measured value, e.g.
scaling, status

AI OUT
VALUE
value status

Temperatur
Sensor Value
Trimmed Value

2ND CYCLIC VALUE

2nd Cyclic
Value
value status

Secondary Value 1

e.g.
PLC

Display Value
value status
P01-Pmx7xxxx-02-xx-xx-xx-000

Fig. 25: The block model indicates what type of data can be transmitted between Cerabar S and the Class 1 master (e.g. PLC) during
cyclic data communication. Using the configuration software of your PLC, compile the cyclic data telegram with the aidof modules (see also "Modules for the cyclic data diagram" in this section). The parameters, written in CAPS, areparameters in the operating program (e.g. FieldCare) which you can use to make settings for the cyclic data telegram or
to display values (see also "Description of parameters" in this section).

Cerabar S function blocks
PROFIBUS uses predefined function blocks to describe the function blocks of a device and to
specify uniform data access.
The following blocks are implemented in Cerabar S:•Physical Block:
The Physical Block contains device-specific features such as the device type, manufacturer,
version etc. as well as functions such as write protection management and ID number
switching•Transducer Block:
The Transducer Block contains all the measuring and device-specific parameters of thedevice. In the Cerabar S Transducer Block, the pressure measuring principle is illustrated
for use as a pressure or level transmitter.
•Analog Input Block (function block):
The Analog Input Block contains the signal processing functions of the measured value
such as scaling, special function calculations, simulation etc.

40

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Description of parameters
Parameter name

Description

OUT VALUE

This parameter shows the digital output value of the Analog Input Block.
Menu path FieldCare:PROFILE VIEW  ANALOG INPUT BLOCK  AI PARAMETER
Menu path onsite display:
GROUP SELECTION OPERATING MENU TRANSMITTER INFO PA DATAPA INPUT VALUEThis value is transmitted from the PLC to Cerabar S. The PA INPUT VALUE can bedisplayed on the onsite display (see also this table, SEL. DISPLAY VAL.).
Menu path FieldCare:PROFILE VIEW PHYSICAL BLOCK PB PARAMETER
Menu path onsite display:
GROUP SELECTION OPERATING MENU TRANSMITTER INFO PA DATA

SEL. DISPLAY VAL.

Use this parameter to specify whether the primary value or a value of the PLC isdisplayed on the onsite display.
Menu path FieldCare:
MANUFACTURER VIEW OPERATING MENU DISPLAY or
PROFILE VIEW PHYSICAL BLOCK PB PARAMETER PROFIBUS PA CONF
Menu path onsite display:
GROUP SELECTION OPERATING MENU TRANSMITTER INFO PA DATA
Options:• Primary value (PV): the primary value is displayed on the onsite display.
• Input value: a value from the PLC is displayed on the onsite display(see this table, PA INPUT VALUE).
Example for the "Input value" option:• Two Cerabar S transmitters measure the pressure drop across a filter. The
differential pressure is registered in the PLC. Use the "Input value" option toassign this calculated value to the onsite display.
Factory setting:• Primary value (PV)2ND CYCLIC VALUE

You can use this parameter to specify which value should be transmitted via thebus as the 2ND CYCLIC VALUE.
Menu path FieldCare:PROFILE VIEW PHYSICAL BLOCK PB PARAMETER PROFIBUS PA CONF
Menu path onsite display:
GROUP SELECTION OPERATING MENU TRANSMITTER INFO PB
PARAMETER PROFIBUS PA CONF
Options:• Temperature• Sensor value: corresponds to the SENSOR PRESSURE parameter• Trimmed value: corresponds to the CORRECTED PRESS. parameter• Secondary value 1: corresponds to the PRESSURE parameter
The SENSOR PRESSURE, CORRECTED PRESSURE and PRESSURE parameters are
displayed in the PROCESS VALUES menu (menu path: MANUFACTOR VIEW 
OPERATING MENU  PROCESSINFO  PROCESS VALUES).
The TEMPERATURE parameter is displayed in the TB PARAMETER menu (menupath: PROFILE VIEW  TRANSDUCER BLOCK  TB PARAMETER)
Factory setting:• Temperature

Endress+Hauser

41

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Modules for the cyclic data diagram
Cerabar S makes the following modules available for the cyclic data diagram:•Main process value
Depending on the operating mode selected, a pressure or level value is transmitted here.
•2ND CYCLIC VALUE
Depending on the option selected, a temperature, the sensor value, trimmed value or
secondary value 1 is transmitted here.
•Display value
This is any value that is transmitted from the PLC to Cerabar S. This value can also be
displayed on the onsite display.
•FREE PLACE
Select this empty module if a value should not be used in the data telegram.
Structure of the output data PLC Cerabar S
With the Data_Exchange service, a PLC can read output data from Cerabar S in the calltelegram. The cyclic data telegram has the following structure:Index
output data

Data

AccessData format/comments

0, 1, 2, 3

Display value

Write

32 bit floating point number (IEEE 754)

4

Status codeWrite

See "Status codes"

Structure of the input data Cerabar S  PLC
With the Data_Exchange service, a PLC can read input data from Cerabar S in the responsetelegram. The cyclic data telegram has the following structure:

42

Index
input data

Data

AccessData format/comments

0, 1, 2, 3

Main process value:pressure or level

Read

32 bit floating point number (IEEE 754)

4

Status code formain process value

Read

See "Status codes"

5, 6, 7. 8

Read
2ND CYCLIC VALUE:
temperature, sensor value,
trimmed value or
secondary value 1

32 bit floating point number (IEEE 754)

9

Status code for
2ND CYCLIC VALUE

See "Status codes"

Read

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Status codes
Cerabar S supports the "Condensed status" function as defined in the PNO specification.
However, the "Classic" status is also supported to ensure compatibility with older devices in
the S series and due to the profile-specific identification number ("0x9700").
If the profile number and new identification number are selected, the status type can be set
via the "COND.STATUS DIAG" parameter.
The "Condensed" and/or "Classic" status and their current active states are displayed by the"Physical Block" in the "Feature" parameter. The measuring device supports the following
status codes for the output value parameters of the Analog Input Block:Classic status:

Endress+Hauser

Statuscode

Device status Meaning

Output value (OUT Value)(Analog Input)

2ND CYCLIC
VALUE

0000
0000

Bad

Not specific

X1)

X

0000
0100

Bad

Configuration error (e.g. adjustment X)not performed correctly)

X

0000
1100

Bad

Device error

X)

X

0001
0000

Bad

Sensor error

X)

X

0001
1100

Bad

Out of service (target mode)

X

X

0100
0000

Uncertain

Not-specific

X

X

0100
0100

Uncertain

Last valid value(failure behavior =1)

X

X

0100
1000

Uncertain

Substitute value (failure behavior =
0)

X

X

0100
1100

Uncertain

Initial value (failure behavior = 1)

X

X

0101
1100

Uncertain

Configuration error (e.g.
linearization table not monotonic
increasing)

X

X

0101
0011

Uncertain

Sensor conversion not accurate constant

X

X

0101
0010

Uncertain

Sensor conversion - limit value
exceeded

X

X

0101
0001

Uncertain

Sensor conversion - limit value
undershot

X

X

0110
0000

Uncertain

Simulation value

X

X

1000
0000

GOOD

Good

X

X

1000
1000

GOOD

Warning limit

X

X

1000
1001

GOOD

Warning limit - limit value exceeded X

X

1000
1010

GOOD

Warning limit - limit value
undershot

X

X

1000
1100

GOOD

Alarm limit

X

X

1000
1101

GOOD

Alarm limit - limit value exceeded

X

X

43

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Statuscode

Device status Meaning

Output value (OUT Value)(Analog Input)

2ND CYCLIC
VALUE

1000
1110

GOOD

X

X

Alarm limit - limit value undershot

1) Only if the analog input failure behavior = 2 ("Status Bad")

Condensed status:
The main reason for implementing the "Condensed" status mode in the Profibus PA Profile
3.02 is to clarify the diagnostic events resulting from use in the PCS/DCS and in theoperating station.
Furthermore, this functionality also implements the requirements of NE 107.
The following "Condensed" status codes are configured via the device.
Statuscode1)

Device status

Meaning

Output value (OUT
2ND CYCLIC
Value) (Analog Input 1) VALUE

0010 01xx

Bad2)

Maintenance alarm, advanced
diagnostics presentX)X

0010 10xx

Bad )

Process error, no maintenancerequired

X3)

X0011 11xx

Bad )

Function check / local override

X)X

Switch offX

X0111 1011 Uncertain

Process error, no maintenancerequired - limit value constantX

X

0111 1010 Uncertain

Process error, no maintenancerequired - limit value exceeded

X

X

0111 1001 Uncertain

Process error, no maintenancerequired - limit value undershot

X

X0111 1000 Uncertain

Process error, no maintenancerequired

X

X

0110 10xx

Maintenance required

X

X

0100 1011 Uncertain

Substitute value

X

0100 1111 Uncertain

Initial valueX0111 0011 Uncertain

Simulated value, startX

0111 0100 Uncertain

Simulated value, end

X1000 0000 GOOD

Good

X

X

1011 1100 GOOD

Function check

X

X

1010 01xx

GOOD

Maintenance required

X

X

1010 10xx

GOOD

Maintenance required

X

X

0010 0011 Bad)Uncertain

X1) Variable x: 0 or 1
2) See  Kap. 9.2.1
3) Only if the analog input failure behavior = 2 ("Status Bad")

44

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.8

Operation

Acyclic data exchange

Acyclic data exchange is used:•To transmit commissioning or maintenance parameters•To display measured variables not contained in the cyclic data diagram.
Using acyclic data exchange, device parameters can be modified even when the device isinvolved in cyclic data exchange with a PLC.
There are two types of acyclic data exchange:•Acyclic communication via the C2 channel (MS2)•Acyclic communication via the C1 channel (MS1)
Acyclic communication via the C2 channel (MS2)
When communicating via the C2 channel, a master opens a communication channel bymeans of a service access point (SAP) to access the device. A master that supports acyclic
communication via the C2 channel is called a Class 2 master. FieldCare is an examples of aClass 2 master.
All the device parameters have to be made known to the master before data can beexchanged via PROFIBUS.
For this, you have the following options:•A configuration program in the master that accesses the parameters via the slot and index
addresses (e.g. FieldCare)•A software component (DTM: Device Type Manager)•The DTM is available on the FieldCare CD.
•The number of Class 2 masters that can simultaneously communicate with a device isrestricted to the number of SAPs available for this communication. Cerabar S supports
MS2 communication with two SAPs. Here, you must make certain that they do not both
attempt to write-access the same data, since otherwise the data consistency cannot be
guaranteed.
•Using the C2 channel for acyclic data exchange increases the cycle times of the bus system.
This should be taken into account when programming the control system.
Acyclic communication via the C1 channel (MS1)
With acyclic communication via the C1 channel, a master that is already communicating
cyclically with the device also opens an acyclic communication channel via SAP 0x33 (special
SAP for MS1). The master can then acyclically read or write the parameters like a Class 2
master via slot and index addresses. Cerabar S supports MS1 communication with one SAP.
NOTICE

Shortening the lifetime of the device!
Parameters written acyclically are written to memory modules (EEPROM, Flash, etc.). These
are resistant to voltage. The memory modules are only designed for a limited number ofwrites which is not even remotely reached in normal operation without MS1 (duringconfiguration). This figure can be quickly exceeded as a result of incorrect programming andthus the operating time of a device can be drastically reduced.
‣In the application program, avoid constantly writing parameters, e.g. for every cycle of the
program.

Endress+Hauser

45

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.9

Slot/index tables

The device parameters are listed in the following tables. You can access the parameters bymeans of the slot and index number. The individual blocks each contain standard
parameters, block parameters and manufacturer-specific parameters.
If you use the FieldCare as an operating program, input screens are available as a userinterface.
General explanatory remarks
Object type
•Record: contains data structure (DS)
•Array: group of a certain data type
•Simple: contains individual data types such as Float
Data type•DS: data structure, contains data types such as Unsigned8, Octet String etc.
•Float: IEEE 754 format•Integer:–Integer8: value range = –128 to 127–Integer16: value range = 327678 to –327678–Integer32: value range =32 = –231 to 231•Octet String: binary coded•Visible String: ASCII coded•Unsigned:–Unsigned8: value range = 0 to 255–Unsigned16: value range = 0 to 65535–Unsigned32: value range = 0 to 4294967295
Storage Class•Cst: constant parameter•D: dynamic parameter•N: nonvolatile parameter•S: static parameter
Device management
Parameter

Slot

Index

Object type Data type

Size (byte)

Storage Class Read

Directory object header
Composite list directory entries
GAP directory continuous
GAP reserved

1
1
1
1

0
1
2–8
9 – 15

Array
Array

12
24

Cst
Cst

46

Unsigned16
Unsigned16

Write

x
x

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Physical Block
Parameter

Slot

Index

Object type Data type

Size (byte)

Storage Class Read

Write

0
0
0
0
0
0
0
0

16
17
18
19
20
21
22
23

RecordSimpleSimpleSimpleSimpleSimpleRecordRecordDS-32
Unsigned16
Visible String
Unsigned16
Unsigned8
Unsigned8
DS-37
DS-42

20
2
32
2
1
1
3
8

Cst
N
S
S
S
S
D
D

x
x
x
x
x
x
x
x

x
x
x
x

SOFTWARE VERSION0HARDWARE REV.
0
MANUFACTOR ID
0
DEVICE NAME STR.
0
DEVICE SERIAL No.
0
DIAGNOSIS0
DIAGNOSIS EXTENSION0DEVICE CERTIFICATION
0INSERT PIN No
0
ADDITIONAL INFO0
USER DESCRIPTION0INSTALLATION DATE0IDENT NUMBER SEL
0
DIP STATUS
0
FEATURE0
COND.STATUS DIAG
0
Physical Block, Endress+Hauser parameters

24
25
26
27
28
29
30
33
34
36
37
38
40
41
42
43

SimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleRecordSimpleVisible String
Visible String
Unsigned16
Visible String
Visible String
Octet String
Octet String
Visible String
Unsigned16
Visible String
Visible String
Visible String
Unsigned8
Unsigned8
DS-68
Unsigned8

16
16
2
16
16
4
6
32
2
32
32
16
1
1
8
1

Cst
Cst
Cst
Cst
Cst
D
D
Cst
N
S
S
S
S
D
N
S

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

ALARM STATUS
LAST DIAG. CODE
UP_DOWN_FEAT
UP/DOWNLOAD CTRL
UP/DOWN PARAM
BUS.ADDRESSSET_UNIT_TO_BUS
PA INPUT VALUE
SEL. DISPLAY VAL.
PROFILE_REVRESET ALL ALARMSIDENT_NUMBER
2ND CYCLIC VALUEDEVICE DESIGN.
CONFIG RECORDEROPERATING HOURS
SIM. ERROR NO.
SIM. MESSAGESLANGUAGEDISPLAY CONTRASTMENU DESCRIPTOR
MAIN DATA FORMATALTERNATE DATAUNIT TEXTUSER DESCRIPTION
ACK. ALARM MODE
ACK. ALARM
SELECT ALARM TYPEERROR NO.
ALARM DELAY
ALARM DISPL. TIME
3RD CYCLIC VALUEHistoROM AVAIL.
HIST. SAVING CYCLHistoROM CONTROLELECTR. SERIAL NO.

54
55
56
57
58
59
61
62
63
64
65
66
68
69
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90
93
94
95
96
97

SimpleSimpleSimpleSimpleSimpleSimpleSimpleRecordSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleUnsigned16
Unsigned16
Unsigned8
Unsigned8
OctetString
Unsigned8
Unsigned8
E+H specificUnsigned8
Visible String
Unsigned8
Unsigned16
Unsigned8
Visible String
Unsigned16
Unsigned32
Unsigned16
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Visible String
Visible String
Unsigned8
Unsigned8
Unsigned8
Unsigned16
Float
Float
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Visible String

2
2
1
1
20
1
1
6
1
32
1
2
1
32
2
4
2
1
1
1
1
1
1
8
32
1
1
1
2
4
4
1
1
1
1
32

D
D
Cst
D
D
D
S
D
S
Cst
S
D
S
S
D
D
D
D
N
S
N
D
N
S
S
S
D
S
D
S
S
S
D
S
S
Cst

x
x
x

Physical Block standard parameters
BLOCK OBJECTSTATIC REVISION NO.
TAG_DESCSTRATEGY
ALERT KEY
TARGET MODE
MODE BLKALARM SUM
Physical Block parameters

Endress+Hauser

0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

x
x
x
x
x

x

x
x
x
x
x
x

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

47

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Parameter

Slot

Index

Object type Data type

Size (byte)

Storage Class Read

PCB TEMPERATURE
Allowed Min. TEMP
Allowed Max. TEMPPCB COUNT: T>Tmax
PCB MAX. TEMP.
PCB COUNT: T < Tmin
PCB MIN. TEMP.
MAIN DATA FORMATDOWNLOAD FUNCT.
STATUS LOCKING
DEVICE STATUS
STATUS SELECT EVENT 727
STATUS SELECT EVENT 115
STATUS SELECT EVENT 120
STATUS SELECT EVENT 731
STATUS SELECT EVENT 730
STATUS SELECT EVENT 733
STATUS SELECT EVENT 732
STATUS SELECT EVENT 726
STATUS SELECT EVENT 715
STATUS SELECT EVENT 720
STATUS SELECT EVENT 717
STATUS SELECT EVENT 718
STATUS SELECT EVENT 740
STATUS SELECT EVENT 716
STATUS SELECT

0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0
0

98
99
100
101
102
103
104
106
107
108
109
110
111
112
113
114
115
116
117
118
119
120
121
122
123
124

SimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleRecord

4
4
4
2
4
4
4
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
14

D
Cst
Cst
D
D
D
D
D
N
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

SWITCH_STATUS_LISTSENSOR SER. No.

0
0

125
126

RecordSimple

Float
Float
Float
Unsigned16
Float
Unsigned16
Float
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
14xUnsigned
82x Unsigned8
Visible String

2
16

D
S

x
x

Write

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

Analog Input Block
Parameter

Slot

Index

Object type

Data type

Size (byte)

Storage Class Read

Write

BLOCK OBJECTSTATIC REVISION NO.
TAG_DESCSTRATEGY
ALERT KEY
TARGET MODE
MODE BLKALARM SUM
Analog Input Block parameters

1
1
1
1
1
1
1
1

16
17
18
19
20
21
22
23

RecordSimpleSimpleSimpleSimpleSimpleRecordRecordDS-32
Unsigned16
Visible String
Unsigned16
Unsigned8
Unsigned8
DS-37
DS-42

20
2
32
2
1
1
3
8

Cst
N
S
S
S
S
D
D

x
x
x
x
x
x
x
x

x
x
x
x

BATCH
OUT
PV SCALEOUT SCALE
LIN TYPECHANNELFILTER TIME CONST
FAIL SAFE MODE
FAIL SAFE DEFAULT VALUE
LIMIT HYSTERESISUPPER LIMIT ALARM
UPPER LIMIT WARNING
LOWER LIMIT WARNING
LOWER LIMIT ALARM
HI_HI_ALM
HI_ALM
LO_ALM
LO_LO_ALARMSIMULATE
VIEW_1_FB

1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1
1

24
26
27
28
29
30
32
33
34
35
37
39
41
43
46
47
48
49
50
61

RecordRecordArray
RecordSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleRecordRecordRecordRecordRecordSimple

DS-67
DS-33
Float
DS-36
Unsigned8
Unsigned16
Float
Unsigned8
Float
Float
Float
Float
Float
Float
DS-39
DS-39
DS-39
DS-39
DS-50
Octet String

10
5
8
11
1
2
4
1
4
4
4
4
4
4
16
16
16
16
6
18

S
D
S
S
S
S
S
S
S
S
S
S
S
S
D
D
D
D
S
D

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

Analog Input Block standard parameters

x
x1)
x
x
x
x
x
x
x
x
x
x
x
x

x

1) If MODE BLK Actual = Manual (MAN)48

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Transducer Block
Parameter

Slot

Index

Object type Data type

Size (byte)

Storage Class Read

2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2

16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
40
41
42
43
44
45
46
47
48
49
50
51
52
53
54
55
56
57
58
59
60
61
62
63
64
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90

RecordSimpleSimpleSimpleSimpleSimpleRecordRecordSimpleSimpleSimpleSimpleSimpleSimpleSimpleRecordSimpleSimpleRecordSimpleSimpleSimpleSimpleSimpleSimpleSimpleRecordSimpleRecordSimpleRecordSimpleSimpleArray
Array
SimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleArray
SimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimple

20
2
32
2
1
1
3
8
4
4
4
4
4
4
2
5
2
4
5
2
2
2
2
2
2
2
5
2
5
2
5
2
1
8
8
4
4
1
1
1
1
1
1
8
4
4
4
4
4
4
2
2
4
4
4
4
4
4
4
4
1
2
2
2

Cst
N
S
S
S
S
D
D
D
N
N
S
S
N
S
D
N
N
D
S
S
S
S
S
S
S
D
S
D
S
D
S
S
S
S
S
S
N
D
N
N
D
D
D
N
N
N
N
S
S
N
N
S
S
S
S
S
S
S
D
D
D
D
D

Write

Transducer Block standard parameters
BLOCK OBJECTSTATIC REVISION NO.
TAG_DESCSTRATEGY
ALERT KEY
TARGET MODE
MODE BLKALARM SUMSENSOR PRESSURE
PRESS.SENS HILIMPRESS.SENS LOLIMHIGH SENSOR TRIMLOW SENSOR TRIM
MINIMUM SPANPRESS. ENG. UNIT
TRIMMED_VALUE (Corrected Press.)
SENSOR MEAS.TYPE
SENSOR SER. No.
PRIMARY VALUE (Measured Value)
PRIM_VALUE_UNIT
PRIM_VALUE_TYPE
MAT. MEMBRANE
FILLING FLUIDSEAL TYPE
PROC.CONN.TYPE
MAT.PROC.CONN. +TB TEMPERATURE (Sensor Temp.)
TEMP. ENG UNIT
SEC_VALUE_1 (PRESSURE)
SEC_VALUE1_UNIT
SEC_VALUE_2
SEC_VALUE2_UNIT
LIN_TYPSCALE_IN
SCALE_OUT
LOW_FLOW_CUT_OFF
FLOW_LIN_SQUARETAB_ACTUAL_NUMB
LINE-NUMB
TAB_MAX_NR
TAB_MIN_NR
TAB_OP_CODE
TAB_STATETAB_XY_VALUE
MAX. MEAS. PRESS.
MIN. MEAS. PRESS.
MAX. MEAS.TEMP.
MIN. MEAS. TEMP.
EMPTY CALIB.
FULL CALIB.
TANK CONTENT UNIT
UNIT FLOWDAMPING VALUEMAX FLOW
MAX PRESS. FLOW
PminALARM WINDOWPmaxALARM WINDOW
TminALARM WINDOW
TmaxALARM WINDOWSIMULATED VALUESIMULATION MODE
COUNTER P>Pmin
COUNTER P<Pmax
COUNTER T>Tmax

Endress+Hauser

DS-32
Unsigned16
Visible String
Unsigned16
Unsigned8
Unsigned8
DS-37
DS-42
Float
Float
Float
Float
Float
Float
Unsigned16
DS-33
Unsigned16
Unsigned32
DS-33
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
DS-33
Unsigned16
DS-33
Unsigned16
DS-33
Unsigned16
Unsigned8
Float
Float
Float
Float
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Float
Float
Float
Float
Float
Float
Float
Unsigned16
Unsigned16
Float
Float
Float
Float
Float
Float
Float
Float
Unsigned8
Unsigned16
Unsigned16
Unsigned16

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

x
x
x
x

x
x

x
x

x
x
x
x
x
x
x
x
x
x
x
x

x
x
x1)
x1
x1
x1
x
x
x
x
x
x
x
x
x
x
x
x

49

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Parameter

Slot

Index

Object type Data type

Size (byte)

Storage Class Read

COUNTER T<Tmin
MEAS. VAL. TRENDTOTALIZER 1
TOTAL. 1 OVERFLOWTOTALIZER 2TOTAL. 2 OVERFLOWTEMP Abs RANGE
Tmin SENSOR
Tmax SENSORSENSOR H/WARE REV.
Pmax PROC. CONN.
TOTAL. 1 ENG. UNITTOTAL. 2 ENG. UNIT
FACT.U.U.TOTAL.1
FACT.U.U.TOTAL.2
TOT. 1 USER UNIT
TOT. 2 USER UNIT
NEG. FLOW TOT. 1
NEG. FLOW TOT. 2
RESET TOTALISER1
FLOW-MEAS. TYPECUSTOMER UNIT F
CUST.UNIT FACT.F
CUSTOMER UNIT P
CUST.UNIT FACT.P
POS.ZERO ADJUSTPOS. INPUT VALUECALIB. OFFSETTANK DESCRIPTION
LIN. EDIT MODE
CALIBRATION MODE
ADJUST DENSITYLEVEL UNIT TXT
CUST.UNIT FACT.L
CUST. UNIT CONT.
FACTOR TANK CONT.
DENSITY UNIT
ADJUST DENSITYTANK VOLUMETANK HEIGHT
100% POINT
ZERO POSITIONLEVEL MIN.
LEVEL MAX.
PROCESS DENSITYMAX TURNDOWNSENSOR CHANGESP PEAKHOLD.STEP
T PEAKHOLD.STEP
ACC. OF GRAVITYCREEP FLOW HYST.
LEVEL BEFORE LIN.
ENG. UNIT LEVELUNIT VOLUMECUSTOMER UNIT V
CUST.UNIT FACT.V
SET.L.FL.CUT-OFF
MAT.PROC.CONN. TANK CONTENT
SUPPRESSED FLOWRESET PEAKHOLDMEASURING MODE
UNIT FLOWTOTALIZER 1 UNIT (Volume p. cond.)
TOTALIZER 2 UNIT (Volume p. cond.)
LOW FLOW CUT-OFF
LO TRIM MEASUREDHI TRIM MEASURED

2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2

91
92
93
94
95
96
97
98
99
100
101
102
103
104
105
106
107
108
109
110
111
112
113
114
115
116
117
118
119
120
121
122
123
124
125
126
127
128
129
130
131
132
133
134
135
136
137
138
139
140
141
142
145
146
147
148
149
150
151
152
153
154
155
156
157
158
159
160

SimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimple

2
1
8
8
8
8
4
4
4
1
4
2
2
4
4
8
8
1
1
1
1
8
4
8
4
1
4
4
32
1
1
4
8
4
8
4
2
4
4
4
4
4
4
4
4
4
2
4
4
4
4
4
2
2
8
4
4
2
4
4
1
1
2
2
2
1
4
4

D
D
D
D
D
D
Cst
Cst
Cst
Cst
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
D
S
S
S
N
S
N
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
S
D
S
S
S
S
S
S
D
D
D
S
S
S
S
S
N
N

50

Unsigned16
Unsigned8
Visible String
Visible String
Visible String
Visible String
Float
Float
Float
Unsigned8
Float
Unsigned16
Unsigned16
Float
Float
Visible String
Visible String
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Visible String
Float
Visible String
Float
Unsigned8
Float
Float
Visible String
Unsigned8
Unsigned8
Float
Visible String
Float
Visible String
Float
Unsigned16
Float
Float
Float
Float
Float
Float
Float
Float
Float
Unsigned16
Float
Float
Float
Float
Float
Unsigned16
Unsigned16
Visible String
Float
Float
Unsigned16
Float
Float
Unsigned8
Unsigned8
Unsigned16
Unsigned16
Unsigned16
Unsigned8
Float
Float

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

Write

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

x
x
x
x
x
x

x
x
x
x
x
x

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Parameter

Slot

Index

Object type Data type

Size (byte)

Storage Class Read

Write

PERCENT UNIT
X-VAL:Y-VAL:MASS FLOW UNIT
SIM. FLOW VALUE
STD. FLOW UNIT
NORM FLOW UNITTOTALIZER 1 UNIT (Mass p. cond.)
TOTALIZER 2 UNIT (Mass p. cond.)
TOTALIZER 1 UNIT (Gas. std. conditions)
TOTALIZER 2 UNIT (Gas. std. conditions)
TOTALIZER 1 UNIT – (Gas. norm conditions)
TOTALIZER 2 UNIT – (Gas. norm conditions)
MASS UNIT
CUST.UNIT FACT.M
CUSTOMER UNIT M
HEIGHT UNIT
CUST.UNIT FACT.H
CUSTOMER UNIT H
EMPTY PRESSURE
FULL PRESSURE
SIM. LEVEL
SIM. TANK CONT.
LEVEL MODE
ACTIV LIN.TAB.X
X-VAL (semi-autom.):
TANK CONTENT MAX.
TANK CONTENT MIN.
HYDR. PRESS MAX.
TAB. ACTIVATETABLE EDITORACTIVE LIN. TAB. Y
HYDR. PRESS MIN.
VALUE LIN. MIN.
VALUE LIN. MAX.
TOTALIZER 1
TOTALIZER 2
LIN. MEASURAND
LINd. MEASURAND
COMB.MEASURAND
TABLE SELECTIONTABLE EDITOR
AREA UNIT
SIM. PRESSURE
PRESSURE ABS RNG
PRESSURE INVERTHEIGHT UNIT
CALIBRATION MODEEMPTY HEIGHTFULL HEIGHTDENSITY UNIT
ADJUST DENSITYPROCESS DENSITY
MEAS.LEVEL EASYLEVEL SELECTIONOUTPUT UNIT

2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2
2

161
162
163
164
165
166
167
168
169
170
171
172
173
174
175
176
177
178
179
180
181
182
183
184
185
186
188
189
190
191
192
193
194
195
196
197
198
199
200
201
202
203
204
205
206
207
240
241
242
243
244
245
246
247
248
249

SimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimpleSimple

2
4
4
2
4
2
2
2
2
2
2
2
2
2
4
8
2
4
8
4
4
4
4
4
4
4
4
4
4
1
1
4
4
4
4
4
4
1
1
1
1
1
2
4
4
1
2
1
4
4
2
4
4
4
1
2

Cst
N
N
S
D
S
S
S
S
S
S
S
S
S
S
S
S
S
S
N
N
D
D
S
N
D
S
S
S
D
N
N
S
S
S
D
D
S
S
S
S
S
S
D
Cst
N
S
S
S
S
S
S
S
N
S
S

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

Unsigned16
Float
Float
Unsigned16
Float
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Float
Visible String
Unsigned16
Float
Visible String
Float
Float
Float
Float
Float
Float
Float
Float
Float
Float
Unsigned8
Unsigned8
Float
Float
Float
Float
Float
Float
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned16
Float
Float
Unsigned8
Unsigend16
Unsigned8
Float
Float
Unsigned16
Float
Float
Float
Unsigned8
Unsigned16

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

x
x
x

x
x
x
x
x
x
x
x

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

1) can only be reset

Endress+Hauser

51

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.3.10

Data format

In the case of PROFIBUS PA, the cyclic transmission of analog values to the PLC is effectedin data blocks 5 bytes long. The measured value is portrayed in the first 4 bytes in the formof floating point numbers in accordance with IEEE standard. The 5th byte contains
standardized status information belonging to the device.
Byte 1

Byte 2Byte 3

Byte 4

Byte 5

Measured value as IEEE 754 floating point number

StatusThe measured value is transmitted as an IEEE 754 floating point number as follows:
Measured value = (–1)sign x 2(E – 127) x (1 + F)
D15

D14

Sign

Exponent (E)
27

D13

D12

D11

D10

D9

D8

D7

D6

D5

D4

D3

D2

D1

D0

Fraction (F)

26

25

24

23

22

21

20

2–1

2–2

2–3

2–4

2–5

2–6

2–7

2–10

2–11

2–12

2–13

2–14

2–15

2–16

2–17

2–18

2–19

2–20

2–21

2–22

2–23

Fraction (F)
2–8

2–9

Example40 F0 00 00 hex = 0100 0000 1111 000 000 000 000 0000 binary
Value = (–1)0 x 2(129 –127) x (1 + 2–1 +2–2 +2–3)= 1 x 22 x (1 + 0.5 +0.25 +0.125)= 1 x 4 x 1.875= 7.5•Not all programmable logic controllers support the IEEE 754-format.
A conversion module must then be used or written.
•Depending on the type of data management (Most-Significant-Byte or Low-SignificantByte) used in the PLC (master), it may also be necessary to change the byte sequence(byte-swapping routine).
Data strings
Some data types, e.g. DS-36, are listed in the slot/index table. These data types are data
strings, structured according to PROFIBUS PA Specification Part 1, Version 3.x.
They consist of several elements that are addressed by means of the slot, index and subindex:

52

Parameter name

Type

Slot

Index

Element

Subindex
1
5

OUT

DS-33

1

26

OUT VALUEOUT STATUS

Parameter name

Type

Slot

Index

Element

OUT SCALE

DS-36

1

28

EU_100_PERCENT
EU_0_PERCENTUNITS_INDEX

Subindex
1
59DECIMAL POINT

11

Type
Float
Unsigned8

Size(byte)
4
1

Type

Size
(byte)
Float
4
Float
4
Unsigned1 2
6Integer8
1

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.4

Operation

Onsite operation – onsite display connected

If the onsite display is connected, the three operating keys are used to navigate through theoperating menu, see  ä 32, "Function of the operating elements – onsite display
connected".

6.4.1

General structure of the operating menu

The menu is split into four levels. The three upper levels are used to navigate while you use
the bottom level to enter numerical values, select options and save settings.
The structure of the OPERATING MENU depends on the measuring mode selected, e.g. if the"Pressure" measuring mode is selected, only the functions necessary for this mode aredisplayed.

Measured value

GROUP SELECTIONLANGUAGE

MEASURING MODEQUICK SETUPOPERATING MENU➁➀SETTINGS➃DISPLAYPOSITION ADJUSTMENT

BASIC SETUP

POS. ZERO ADJUST➂EXTENDED SETUP

POS. INPUT VALUECALIB. OFFSET

P01-xxxxxxxx-19-xx-xx-xx-097

Fig. 26: General structure of the operating menu
11. Selection level
22. Selection level
3Function groups
4Parameter

The LANGUAGE and MEASURING MODE parameters are only displayed via the onsite
display on the 1st selection level. In FieldCare, the LANGUAGE parameter is displayed in the
DISPLAY group and the MEASURING MODE parameter is displayed in the QUICK SETUP
menus or in the BASIC SETUP function group.

Endress+Hauser

53

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.4.2

Selecting an option

Example: select "English" as the language of the menu.
Onsite display

Operation
German is selected as the language. A 3 in front of themenu text indicates the active option.

P01-xxxxxxxx-19-xx-xx-xx-017

Select English with "+" or "–".

P01-xxxxxxxx-19-xx-xx-xx-033

1. Confirm your choice with "E". A 3 in front of themenu text indicates the active option.
(English is now selected as the menu language.)
2. Jump to the next item with "E".
P01-xxxxxxxx-19-xx-xx-xx-034

6.4.3

Taking pressure applied at device as value

Example: performing position adjustment.
Onsite display

Operation
The bottom line on the onsite display displays thepressure present, here 3.9 mbar.

P01-xxxxxxxx-19-xx-xx-xx-158

Use "+" or "–" to switch to the "Confirm" option. The
active selection is highlighted in black.

P01-xxxxxxxx-19-xx-xx-xx-159

Use "E" to assign the value (3.9 mbar) to the POS. ZEROADJUST parameter. The device confirms the calibrationand jumps back to the parameter, here POS. ZEROADJUST (see next graphic).
P01-xxxxxxxx-19-xx-xx-xx-037

Switch to the next parameter with "E".

P01-xxxxxxxx-19-xx-xx-xx-160

54

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.4.4

Operation

Editing a value

Example: adjusting DAMPING VALUE function from 2.0 s to 30.0 s.
See also  ä 32, "Function of the operating elements – onsite display connected".
Onsite display

Operation
The onsite display shows the parameter to be changed.
The value highlighted in black can be changed. The "s"unit is fixed and cannot be changed.

P01-xxxxxxxx-19-xx-xx-xx-023

1. Press "+" or "–" to get to the editing mode.
2. The first digit is highlighted in black.

P01-xxxxxxxx-19-xx-xx-xx-027

1. Use "+" to change "2" to "3".
2. Confirm "3" with "E". The cursor jumps to the next
position (highlighted in black).

P01-xxxxxxxx-19-xx-xx-xx-028

The decimal point is highlighted in black, i.e. you can
now edit it.

P01-xxxxxxxx-19-xx-xx-xx-029

1.

Keep pressing "+" or "–" until "0" is displayed.

2. Confirm "0" with "E".
The cursor jumps to the next position.  isdisplayed and is highlighted in black. See nextgraphic.
P01-xxxxxxxx-19-xx-xx-xx-030

Use "E" to save the new value and exit the editing mode.
See next graphic.

P01-xxxxxxxx-19-xx-xx-xx-031

The new value for the damping is now 30.0 s.
– Jump to the next parameter with "E".
– You can get back to the editing mode with "+" or "–".

P01-xxxxxxxx-19-xx-xx-xx-032

Endress+Hauser

55

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.5

Endress+Hauser operating program

The FieldCare operating program is an Endress+Hauser asset management tool based on
FDT technology. With FieldCare, you can configure all Endress+Hauser devices as well as
devices from other manufacturers that support the FDT standard. Hardware and software
requirements you can find on the internet: www.endress.com  select your country Search: FieldCare  FieldCare Technical Data.
FieldCare supports the following functions:•Configuration of transmitters in online operation•Loading and saving device data (upload/download)•Tank linearization•HistoROM®/M-DAT analysis•Documentation of the measuring point
Connection options:•PROFIBUS PA via segment coupler and PROFIBUS interface card
•PROFIBUS PA via Fieldgate FXA720, segment coupler and PROFIBUS interface card
•Commubox FXA291 with ToF Adapter FXA291 via service interface•In the "Level Standard" measuring mode, the configuration data that were loaded with FDT
upload cannot be written again (FDT download). These data are only used to document themeasuring point.
•Further information on the FieldCare can be found on the Internet:
http://www.endress.com  Download  Text Search: FieldCare).

6.6

HistoROM®/M-DAT (optional)

NOTICE

Device could be destroyed!
Detach HistoROM®/M-DAT from the electronic insert or attach it to the insert in adeenergized state only.
HistoROM®/M-DAT is a memory module, which is attached to the electronic insert andfulfils the following functions:•Back-up copy of configuration data
•Copying configuration data of a transmitter into another transmitter•Cyclic recording of pressure and sensor-temperature measured values•Recording diverse events, such as alarms, configuration changes, counters for measuring
range undershooting and exceeding for pressure and temperature, exceeding andundershooting the user limits for pressure and temperature, etc.
•The HistoROM®/M-DAT module may be retrofitted at any time (Order No.: 52027785).
•The HistoROM data and the data in the device are analyzed once a HistoROM®/M-DAT is
attached to the electronic insert and power is reestablished to the device. During theanalysis, the messages "W702, HistoROM data not consistent" and "W706, Configurationin HistoROM and device not identical" can occur. For measures see  ä 75, "Messages."

56

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.6.1

Operation

Copying configuration data

➁
on

t

off

0%

➀

Zero
Display

2

on

on

off

off

ON

1

2

3

CK S D A 0

8

5

8

4

6

7

PC

1

1 2 3 4 5 6 7 8
Address

HW-Version:
SW-Version:

1
2

250002272-–

SW
HW

Histo ROM

Sensor

P01-xxxxxxxx-19-xx-xx-xx-110

Electronic insert with optional HistoROM®/M-DAT memory module
1Optional HistoROM®/M-DAT
2To copy configuration data from the HistoROM®/M-DAT to a device or from a device to a HistoROM®/M-DAT , the operation must
be unlocked (DIP switch 1, position "Off" , parameter INSERT PIN No = 2457). See also  ä 59, "Locking/unlocking operation".

Onsite operation via onsite display (optional) or remote operation
Copying configuration data from a device to a HistoROM®/M-DAT:The operation must be unlocked.
1.Disconnect device from supply voltage.
2.Attach the HistoROM®/M-DAT module to the electronic insert.
3.Reestablish supply voltage to the device.
4.The DOWNLOAD SELECT. parameter (OPERATION menu) setting has no influence on an
upload from the device into HistoROM.
5.Using the HistoROM CONTROL parameter (OPERATION menu) select the option "Device
HistoROM" as the data transfer direction.
6.Wait approx. 20 seconds. Configuration data are loaded from the device to theHistoROM®/M-DAT. The device is not restarted.
7.Disconnect device from the supply voltage again.
8.Detach memory module.
9.Reestablish supply voltage to the device.

Copying configuration data from a HistoROM®/M-DAT to a device:
Endress+Hauser

57

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

The operation must be unlocked.
1.Disconnect device from supply voltage.
2.Attach the HistoROM®/M-DAT module to the electronic insert. Configuration data fromanother device are stored in the HistoROM®/M-DAT.
3.Reestablish supply voltage to the device.
4.Use the DOWNLOAD SELECT parameter (OPERATION menu) to select which parameters
are to be overwritten.
The following parameters are overwritten according to the selection:–Configuration copy (factory setting):
all parameters except DEVICE SERIAL No., DEVICE DESIGN, TAG, DESCRIPTION,
IDENT_NUMBER_SEL, BUS ADDRESS and the parameters in the POSITION
ADJUSTMENT, PROCESS CONNECTION, SENSOR TRIM and SENSOR DATA group.
–Device replacement:
all parameters except DEVICE SERIAL No., IDENT_NUMBER_SEL, DEVICE DESIGN
and the parameters in the POSITION ADJUSTMENT, PROCESS CONNECTION,SENSOR TRIM and SENSOR DATA group.
–Electronics replace:all parameters except the parameters in the SENSOR DATA group.
Factory setting: Configuration copy
5.Using the HistoROM CONTROL parameter select the option "HistoROM Device" as thedata transfer direction.
6.Wait approx. 45 seconds. Configuration data are loaded from the device to theHistoROM®/M-DAT. The device is restarted.
7.Before removing the HistoROM®/M-DAT again from the electronic insert, disconnect thedevice from supply voltage.

58

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.7

Operation

Locking/unlocking operation

Once you have entered all the parameters, you can lock your entries against unauthorized
and undesired access.
You have the following possibilities for locking/unlocking the operation:•Via a DIP-switch on the electronic insert, locally on the display.
•Via the onsite display (optional)•Via communication e.g. FieldCare.
The -symbol on the onsite display indicates that operation is locked. Parameters whichrefer to how the display appears, e.g. LANGUAGE and DISPLAY CONTRAST can still be
altered.

If operation is locked by means of the DIP-switch, you can only unlock operation again bymeans of the DIP-switch. If operation is locked by means of remote operation e.g. FieldCare,you can only unlock operation again by means of remote operation.
The table provides an overview of the locking functions:
View/readparameter

Modify/write via1)

Unlocking via

Onsitedisplay

Remoteoperation

DIP-switch

Onsitedisplay

Remoteoperation

DIP-switch

Yes

No

No

Yes

No

No

Onsite display

Yes

No

No

No

Yes

Yes

Remote operation

Yes

No

No

No

Yes

Yes

Locking via

1) Parameters which refer to how the display appears, e.g. LANGUAGE and DISPLAY CONTRAST can still be
altered.

➁

➂
on

0%
Zero

on

2

off

1

2

off

2

on

on

off

off

ON

1

2

3

4

CK S D A 0

8

5

8

6

7

PC

1

1

Display

Sensor

1 2 3 4 5 6 7 8
Address

HW-Version:
SW-Version:

–

+

1
2

250002272-–

SW
HW

Histo ROM➀

Damping [t]Locking/unlocking operation locally via DIP-switch
Damping [t]

6.7.1

E

P01-xxxxxxxx-19-xx-xx-xx-111

Fig. 27: DIP-switch position "Hardware locking" on the electronic insert
1If necessary, remove onsite display (optional)
2DIP-switch is at "On": operation is locked.
3DIP-switch is at "Off": operation is unlocked (operation possible)

Endress+Hauser

59

Operation

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

6.7.2

Locking/unlocking operation via remote operation
Description

Locking operation

1. Select INSERT PIN No parameter, menu path onsite display:
GROUP SELECTION  OPERATING MENU  OPERATING  INSERT PIN No.
menu path FieldCare:
MANUFACTOR VIEW  OPERATING MENU  OPERATION  INSERT PIN No.
2. To lock operation, enter "0" for the parameter.

Unlocking operation

1. Select INSERT PIN No. parameter.
2. To unlock operation, enter "2457" for the parameter.

6.8

Factory setting (reset)•Total reset: Press zero-key for at least 12 seconds. If the LED on the electronic insert lights
up briefly, the reset is being carried out.
•By entering a certain code, you can completely, or partially, reset the entries for theparameters to the factory settings. (For the factory settings see Operating Instructions
BA00296P "Cerabar S/Deltabar S/Deltapilot S, Description of Device Functions".)
Enter the code by means of the ENTER RESET CODE parameter (OPERATION menu).
There are various reset codes for the device. The following table illustrates whichparameters are reset by the particular reset codes. Operation must be unlocked to resetparameters (see  ä 59, "Locking/unlocking operation").

•Any customer-specific configuration carried out by the factory is not affected by a reset(customer-specific configuration remains). If, after a reset, you wish the parameters to bereset to the factory settings, please contact Endress+Hauser Service.
•The OUT value may have to be rescaled after resetting with code 1, 40864 or 33333.
See also  ä 72, "Scaling the OUT value" and  ä 73, "System units (SET UNIT TO BUS)".

60

Reset code

Description and effect

1 or 40864

Total reset– This reset resets the following parameters:– POSITION ADJUSTMENT function group– BASIC SETUP function group– EXTENDED SETUP function group– LINEARISATION function group (an existing linearization table is deleted)– TOTALIZER SETUP function group– OUTPUT group– PA DATA function group, SET UNIT TO BUS, 2ND CYCLIC VALUE, SEL.DISPLAYVALUE parameters– TRANSMITTER DATA function group, TAG, ADDITIONAL INFO. parameters– MESSAGES function group– All configurable messages ("Error" type) are set to "Warning".
See also  ä 75, "Messages" and  ä 83, "Response of outputs to errors".
– USER LIMITS function group– The bus address is not affected.
– Any simulation which may be running is ended.
– The device is restarted.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Operation

Reset code

Description and effect33333

User reset– This reset resets the following parameters:– POSITION ADJUSTMENT function group– BASIC SETUP function group, apart from customer-specific units– EXTENDED SETUP function group– TOTALIZER SETUP function group– OUTPUT group– PA DATA function group, SET UNIT TO BUS, 2ND CYCLIC VALUE, SEL.DISPLAYVALUE parameters– TRANSMITTER DATA function group, TAG, ADDITIONAL INFO. parameters– Any simulation which may be running is ended.
– The device is restarted.

35710

Reset Level measuring mode
– Depending on the settings for the LEVEL MODE, LIN MEASURAND, LINd
MEASURAND or COMB. MEASURAND parameters, the parameters needed forthis measuring task will be reset.
– Any simulation which may be running is ended.
– The device is restarted.
Example LEVEL MODE = linear and LIN. MEASURAND = level• HEIGHT UNIT = m• CALIBRATION MODE = wet• EMPTY CALIB. = 0• FULL CALIB. = Sensor end value converted to mH2O,
e.g. 5.99 mH2O for a 500 mbar (7.5 psi) sensor

Endress+Hauser

34846

Display reset– This reset resets all parameters which have to do with how the display appears(DISPLAY group).
– Any simulation which may be running is ended.
– The device is restarted.

41888

HistoROM reset
The measured value and event buffers are deleted. During the reset, the HistoROM
has to be attached to the electronic insert.

2506

PowerUp reset (warm start)– This reset resets all the parameters in the RAM. Data are read back anew from the
EEPROM (processor is initialized again).
– Any simulation which may be running is ended.
– The device is restarted.

2712

Bus address reset– The device address configured via the bus is reset to the 126 factory setting.
– Any simulation which may be running is ended.
– The device is restarted.

61

Commissioning

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7

Commissioning

The device is configured for the Pressure measuring mode at the factory. The measuring
range and the unit in which the measured value is transmitted, as well as the digital output
value of the Analog Input Block OUT, correspond to the data on the nameplate. Following a
reset with code 1, 40864 or 33333 the OUT Value may have to be rescaled (see also ä 72, "Scaling the OUT value" and  ä 73, "System units (SET UNIT TO BUS)").
!

WARNING

Exceeding the maximum allowable working pressure!
Risk of injury due to bursting of parts! Warning messages are generated if pressure is too
high.
‣If a pressure greater than the maximum permitted pressure is present at the device, themessages "E115 Sensor overpressure" and "E727 Sensor pressure error - overrange" areoutput in succession! Use the device only within the sensor range limits
NOTICE

Shortfall of the allowable working pressure!
Output of messages if pressure is too low.
‣If a pressure smaller than the minimum permitted pressure is present at the device, themessages "E120 Sensor low pressure" and "E727 Sensor pressure error - overrange" areoutput in succession! Use the device only within the sensor range limits

7.1

Configuring messages•Messages E727, E115 and E120 are "Error"-type messages and can be configured as a"Warning" or an "Alarm". These messages are configured as "Warning" messages at thefactory. This setting prevents the current output from assuming the set alarm current
value for applications (e.g. cascade measurement) where the user is consciously aware of
the fact that the sensor range can be exceeded•We recommend setting messages E727, E115 and E120 to "Alarm" in the following
instances:–The sensor range does not have to be exceeded for the measuring application.
–Position adjustment has to be carried out that has to correct a large measured error as aresult of the orientation of the device (e.g. devices with a diaphragm seal).

7.2

Function check

Carry out a post-installation and a post-connection check as per the checklist before
commissioning the device.
•"Post-installation check" checklist,  ä 25.
•"Post-connection check" checklist,  ä 28.

62

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.3

Commissioning

Commissioning via Class 2 master (FieldCare)

Commissioning and operating of FieldCare are described in the integrated FieldCare online
help.
Proceed as follows to commission the device:
1.Check the hardware write protection on the electronic insert (see  ä 59, "Locking/unlocking operation").
The DIP STATUS parameter shows the status of the hardware write protection (menupath: MANUFACTURER VIEW TRANSMITTER INFO TRANSMITTER DATA)
2.Enter the tag name by means of the ADDITIONAL INFO parameter. (Menu path:
MANUFACTURER VIEW  TRANSMITTER INFO TRANSMITTER DATA or PROFILE
VIEW PB PARAMETER DEVICE)
3.Assign the device an address in the bus (see also  ä 36, "Device identification andaddressing")
4.Configure manufacturer-specific device parameters via the MANUFACTOR VIEW menu.
5.Configure the PHYSICAL BLOCK (menu path: PROFILE VIEW PHYSICAL BLOCK)
6.Configure the ANALOG INPUT BLOCK.
–In the Analog Input Block, the input value or input range can be scaled in accordancewith the requirements of the automation system (see  ä 72, "Scaling the OUT
value") or perform SET.UNIT.TO.BUS (Kap. 7.9).
–If necessary, configure the limit values.
7.Configure the cyclic data communication (see  ä 38, "System integration" and  ä 40,"Cyclic data exchange").

7.4

Selecting language and measuring mode

7.4.1

Local operation

The LANGUAGE and MEASURING MODE parameters are located on the top menu level.
See also  ä 53, "General structure of the operating menu".
The following languages are available:•Deutsch•English•Français
•Italiano•Español
•Nederlands•Chinese (CHS)•Japanese (JPN)The following measuring modes are available:•Pressure•LevelEndress+Hauser

63Commissioning

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.4.2

FieldCare

The MEASURING MODE parameter is displayed in the QUICK SETUP menus and in theBASIC SETUP function group (OPERATING MENU  SETTINGS  BASIC SETUP).
The following measuring modes are available:•Pressure•LevelThe LANGUAGE parameter is arranged in the DISPLAY group (OPERATING MENU DISPLAY).
•Use the LANGUAGE parameter to select the menu language for the onsite display.
•Select the menu language for FieldCare by means of the "Language Button" in theconfiguration window. Select the menu language for the FieldCare frame via the "Extra"
menu  "Options"  "Display"  "Language".
The following languages are available:•Deutsch•English•Français
•Italiano•Español
•Nederlands•Chinese (CHS)•Japanese (JPN)64

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.5

Commissioning

Position adjustment

Due to the orientation of the device, there may be a shift in the measured value, i.e. when
the container is empty or partly filled, the measured value parameter does not display zero.
There are three options to choose from when performing position adjustment.
•Menu path onsite display: GROUP SELECTION  OPERATING MENU  SETTINGS POSITION ADJUSTMENT•Menu path FieldCare: MANUFACTOR VIEW  OPERATING MENU  SETTINGS POSITION ADJUSTMENT
Parameter name

Description

POS. ZERO ADJUST
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure need not be known.
Example:– MEASURED VALUE = 2.2 mbar (0.032 psi)
– Correct the MEASURED VALUE via the POS. ZERO ADJUST parameter with the"Confirm" option. This means that you are assigning the value 0.0 to the pressure
present.
– MEASURED VALUE (after pos. zero adjust) = 0.0 mbar
The CALIB. OFFSET parameter displays the resulting pressure difference (offset) bywhich the MEASURED VALUE was corrected.
Factory setting:
0.0

POS. INPUT VALUE
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure need not be known. To correct the pressure difference, youneed a reference measurement value (e. g. from a reference device).
Example:– MEASURED VALUE = 0.5 mbar (0.0073 psi)
– For the POS. INPUT VALUE parameter, specify the desired set point for the
MEASURED VALUE, e.g. 2.0 mbar (0.029 psi).
(MEASURED VALUEnew = POS. INPUT VALUE)– MEASURED VALUE (after entry for POS. INPUT VALUE) = 2.0 mbar (0.029 psi)
– The CALIB. OFFSET parameter displays the resulting pressure difference (offset)
by which the MEASURED VALUE was corrected.
CALIB. OFFSET = MEASURED VALUEold – POS. INPUT VALUE,
here: CALIB. OFFSET = 0.5 mbar (0.0073 psi) – 2.0 mbar (0.029 psi) = – 1.5
mbar (0.022 psi)
Factory setting:
0.0

CALIB. OFFSET
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure is known. (A reference pressure is not present at the device.)
Example:– MEASURED VALUE = 2.2 mbar (0.032 psi)
– Via the CALIB. OFFSET parameter, enter the value by which the MEASUREDVALUE should be corrected. To correct the MEASURED VALUE to 0.0 mbar, youmust enter the value 2.2 here.
(MEASURED VALUE new = MEASURED VALUEold – CALIB. OFFSET)– MEASURED VALUE (after entry for calib. offset) = 0.0 mbar
Factory setting:
0.0

Endress+Hauser

65Commissioning

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.6

Pressure measurement

7.6.1

Information on pressure measurement•The Level and Pressure operating modes each have a quick setup menu which guides you
through the most important basic functions. With the setting in the MEASURING MODE
parameter, you specify which Quick Setup menu should be displayed. See also  ä 63,"Selecting language and measuring mode".
•For a detailed description of the parameters see the Operating Instructions BA00296P"Cerabar S/Deltabar S/Deltapilot S, Description of device functions"–Table 6, POSITION ADJUSTMENT–Table 7, BASIC SETUP–Table 16, EXTENDED SETUP•For differential pressure measurement, select the "Pressure" option by means of the
MEASURING MODE parameter. The operating menu is structured appropriately.
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣If the measuring mode is changed, the span setting (URV) must be verified in the"Calibration"  "Basic Setup" operating menu and, if necessary, reconfigured!

7.6.2

Quick Setup menu for the Pressure measuring mode
MEASURED VALUE

1)
GROUP SELECTION

1)LANGUAGE

1)
MEASURING MODE

2)QUICK SETUP

MANUFACTOR VIEW

2)
MEASURING MODE

2)
PROFILE VIEW

1)
OPERATING MENU

Pressure

Level
Absolut pressure sensors

Gauge pressure
sensors
POS. ZERO ADJUSTPOS. INPUT VALUE

1) On-site display only
2) FieldCare only

DAMPING VALUE
P01-PMx7xxxx-19-xx-xx-xx-076

Fig. 28: Quick Setup menu for the "Pressure" measuring mode

Local operation

FieldCare

Measured value display
Onsite display: Switch from the measured value
display to GROUP SELECTION with F.

Measured value display
Select QUICK SETUP menu.

GROUP SELECTION
Select MEASURING MODE.

MEASURING MODE
Select "Pressure" option.

MEASURING MODE
Select "Pressure" option.
GROUP SELECTION
Select QUICK SETUP menu.
POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i. e. you assign the value 0.0 to
the pressure present.

66

POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i. e. you assign the value 0.0 to
the pressure present.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Commissioning

Local operation

FieldCare

POS. INPUT VALUE
Due to orientation of the device, there may be a shift
in the measured value. For the POS. INPUT VALUE
parameter, specify the desired set point for the
MEASURED VALUE.

POS. INPUT VALUE
Due to orientation of the device, there may be a shift
in the measured value. For the POS. INPUT VALUE
parameter, specify the desired set point for the
MEASURED VALUE.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
value of the Analog Input Block react to a change inthe pressure.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
value of the Analog Input Block react to a change inthe pressure.

For onsite operation, see also ä 32, "Function of the operating elements – onsite display connected" and ä 53, "Onsite operation – onsite display connected".

Endress+Hauser

67Commissioning

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.7

Level measurement

7.7.1

Information on level measurement•The Level and Pressure operating modes each have a quick setup menu which guides you
through the most important basic functions. See  ä 70 for the "Level" quick setup menu.
•Furthermore, the three level modes "Level Easy Pressure", "Level Easy Height" and "Level
Standard" are available to you for level measurement. You can select from the "Linear","Pressure linearized" and "Height linearized" level types for the "Level Standard" level mode.
The table in the "Overview of level measurement" section below provides an overview of thevarious measuring tasks.
–In the "Level Easy Pressure" and "Level Easy Height" level modes, the values entered are
not tested as extensively as in the "Level Standard" level mode. The values entered for
EMPTY CALIB./FULL CALIB., EMPTY PRESSURE/FULL PRESSURE and EMPTY HEIGHT/
FULL must have a minimum interval of 1% for the "Level Easy Pressure" and "Level Easy
Height" level modes. The value will be rejected with a warning message if the values are
too close together. Further limit values are not checked; i.e. the values entered must be
appropriate for the sensor and the measuring task so that the measuring device canmeasure correctly.
–The "Level Easy Pressure" and "Level Easy Height" level modes encompass fewer
parameters than the "Level Standard" mode and are used for quick and easy
configuration of a level application.
–Customer-specific units of fill level, volume and mass or a linearization table may only beentered in the "Level Standard" level mode.
•See the Operating Instructions BA00296P "Cerabar S/Deltabar S/Deltapilot S, Descriptionof device functions".
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣If the measuring mode is changed, the span setting (URV) must be verified in the"Calibration"  "Basic Setup" operating menu and, if necessary, reconfigured!

68

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.7.2

Commissioning

Overview of level measurement

Measuring task

LEVEL SELECTION/LEVEL MODE

Measured
Description
variable options

The measured variable is in
direct proportion to themeasured pressure.
Calibration is performed byentering two pressure-levelvalue pairs.

LEVEL SELECTION:
Level Easy Pressure

Via OUTPUT
UNIT parameter:%, level, volume
or mass units.

– Incorrect entries– Calibration withare possiblereference pressure – wet– Customized units
calibration, see
are not possible
Operating Instructions
BA00296P.
– Calibration without
reference pressure – drycalibration, see
Operating Instructions
BA00296P.

The measured value
display and the LEVELBEFORE LIN parametershow the measured
value.

The measured variable is in
direct proportion to themeasured pressure.
Calibration is performed byentering the density and twoheight-level value pairs.

LEVEL SELECTION:
Level Easy HeightVia OUTPUT
UNIT parameter:%, level, volume
or mass units.

– Incorrect entries– Calibration withare possiblereference pressure – wet– Customized units
calibration, see
are not possible
Operating Instructions
BA00296P.
– Calibration without
reference pressure – drycalibration, see
Operating Instructions
BA00296P.

The measured value
display and the LEVELBEFORE LIN parametershow the measured
value.

The measured variable is in
direct proportion to themeasured pressure.

LEVEL SELECTION:
Level standard/
LEVEL MODE:
LinearVia LIN.
MEASURAND
parameter:– % (level)– Level– Volume– Mass– Incorrect entries– Calibration withare rejected byreference pressure – wet
the device
calibration, see– Customized level,
Operating Instructions
volume and mass
BA00296P.
units are possible– Calibration without
reference pressure – drycalibration, see
Operating Instructions
BA00296P.

The measured value
display and the LEVELBEFORE LIN parametershow the measured
value.

The measured variable is not in
direct proportion to themeasured pressure as, forexample, with containers witha conical outlet. A linearizationtable must be entered for thecalibration.

LEVEL SELECTION:
Level standard/
LEVEL MODE:
Pressure linearized

Via LINd.
MEASURAND
parameter:– Pressure + %– Pressure +volume– Pressure +mass– Calibration withreference pressure:semiautomatic entry of
linearization table, see
Operating Instructions
BA00296P.
– Calibration without
reference pressure:manual entry of
linearization table, see
Operating Instructions
BA00296P.

– Incorrect entries
are rejected by
the device– Customized level,
volume and massunits are possible

The measured value
display and the
TANK CONTENT
parameter show themeasured value.

– Two measured variables are
required or– The container shape is givenby value pairs, such asheight and volume.

LEVEL SELECTION:
Level standard/
LEVEL MODE:
Height linearized

Via COMB.
MEASURAND
parameter:– Height +volume– Height + mass
– Height + %– %-Height +volume– %-Height +mass– %-Height + %– Calibration withreference pressure: wet
calibration andsemiautomatic entry of
linearization table, see
Operating Instructions
BA00296P.
– Calibration without
reference pressure: drycalibration and manualentry of linearizationtable, see Operating
Instructions BA00296P.

– Incorrect entries
are rejected by
the device– Customized level,
volume and massunits are possible

The measured value
display and the
TANK CONTENT
parameter show the
2nd measured value(volume, mass or %).

The 1st measured variable %height or height must be indirect proportion to themeasured pressure. The 2nd
measured variable volume,
mass or % must not be in direct
proportion to the measured
pressure. A linearization tablemust be entered for the 2nd
measured variable. The 2nd
measured variable is assigned
to the 1st measured variable bymeans of this table.

Endress+Hauser

Comment

Measured value
display

The LEVEL BEFORE
LIN parameter displays
the 1st measured value(%-height or height).

69

Commissioning

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.7.3

Quick Setup menu for the Level measuring mode

•Some parameters are only displayed if other parameters are appropriately configured. For
example, the EMPTY CALIB. parameter is only displayed in the following cases:–LEVEL SELECTION "Level Easy Pressure" and CALIBRATION MODE "Wet"–LEVEL SELECTION "Level Standard", LEVEL MODE "Linear" andCALIBRATION MODE "WET"
You can find the LEVEL MODE parameter in the BASIC SETTINGS function group.
•The following parameters are set to the following values in the factory:–LEVEL SELECTION: Level Easy Pressure–CALIBRATION MODE: Wet–OUTPUT UNIT or LIN. MEASURAND: %–EMPTY CALIB.: 0.0–FULL CALIB.: 100.0•The quick setup is suitable for simple and quick commissioning. If you wish to make more
complex settings, e.g. change the unit from "%" to "m", you will have to calibrate using theBASIC SETTINGS group. See Operating Instructions B00A296P.
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣If the measuring mode is changed, the span setting (URV) must be verified in the"Calibration"  "Basic Setup" operating menu and, if necessary, reconfigured!

MEASURED VALUE

1)
GROUP SELECTION

1)LANGUAGE

2)
MEASURING MODEQUICK SETUP

MANUFACTOR VIEWLEVEL SELECTION

2)
PROFILE VIEWOPERATING MENU

Pressure

Level

Gauge pressure sensors

Absolut pressure sensors

POS. ZERO ADJUSTPOS. INPUT VALUE

3)1) On-site display only

EMPTY CALIB.

2) FieldCare only3) – LEVEL SELECTION "Level Easy Pressure"and CALIBRATION MODE "Wet"– LEVEL SELECTION = "Level Standard",
LEVEL MODE = "Linear" andCALIBRATION MODE = "Wet"

3)
FULL CALIB.

DAMPING VALUE

P01-PMx7xxxx-19-xx-xx-xx-077

Fig. 29: Quick Setup menu for the "Level" measuring mode

Local operation

FieldCare

Measured value display
Onsite display: Switch from the measured value
display to GROUP SELECTION with F.

Measured value display
Select QUICK SETUP menu.

GROUP SELECTION
Select MEASURING MODE.

MEASURING MODE
Select "Level" option.

MEASURING MODE
Select "Level" option.
LEVEL SELECTION
Select level mode. For an overview see  ä 69.

70

LEVEL SELECTION
Select level mode. For an overview see  ä 69.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Local operation

Commissioning

FieldCareGROUP SELECTION
Select QUICK SETUP menu.
POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i. e. you assign the value 0.0 to
the pressure present.

POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i. e. you assign the value 0.0 to
the pressure present.

POS. INPUT VALUE
Due to orientation of the device, there may be a shift
in the measured value. For the POS. INPUT VALUE
parameter, specify the desired set point for the
MEASURED VALUE.

POS. INPUT VALUE
Due to orientation of the device, there may be a shift
in the measured value. For the POS. INPUT VALUE
parameter, specify the desired set point for the
MEASURED VALUE.

EMPTY CALIB. 1)
Enter level for the lower calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

EMPTY CALIB. 1
Enter level for the lower calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

FULL CALIB. 1
Enter level for the upper calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

FULL CALIB. 1
Enter level for the upper calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
value of the Analog Input Block react to a change inthe pressure.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
value of the Analog Input Block react to a change inthe pressure.

1) – LEVEL SELECTION "Level Easy Pressure" and CALIBRATION MODE "Wet"– LEVEL SELECTION "Level Standard", LEVEL MODE "Linear" and CALIBRATION MODE "Wet"

For onsite operation, see also ä 32, "Function of the operating elements – onsite display connected" and ä 53, "Onsite operation – onsite display connected".

Endress+Hauser

71Commissioning

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.8

Scaling the OUT value

In the Analog Input Block, the input value or input range can be scaled in accordance withthe automation requirements.
Example:The measuring range 0 to 500 mbar (0 to 7.5 psi) should be rescaled to 0 to 10000.
•Select PV SCALE group.
Menu path: PROFILE VIEW  ANALOG INPUT BLOCK  AI parameter–Enter "0" as the LOWER VALUE.
–Enter "500" as the UPPER VALUE.
•Select OUT SCALE group.
Menu path: PROFILE VIEW  ANALOG INPUT BLOCK  AI parameter–Enter "0" as the LOWER VALUE.
–Enter "10000" as the UPPER VALUE.
–For UNIT, select "User unit" for example.
The unit selected here does not have any effect on the scaling.
•Result:
At a pressure of 350 mbar (5.25 psi), the value 7000 is output to the PLC as the OUT value.

Analog Input Block
PV SCALE

UPPER VALUE
500 mbar

OUT SCALE

1

MEASURED
VALUE= 350 mbar

0.7

OutputTransducer
Block

LOWER VALUE
0 mbar

0
0

0.7

1

0

affects the bargraph

UPPER VALUELOWER VALUE

10000

0.7 10000

OUT Value, AI Blockto PLC, here for example
10000 • 0,7 (7000)
P01-xMx7xxxx-05-xx-xx-xx-002•The OUT Value can only be scaled via remote operation (e.g. FieldCare).
•When a unit changes within a measuring mode, the limits for PV SCALE are converted.
•If the operating mode is changed, no conversion takes place. The device must berecalibrated following a change in the operating mode.
•Via the "SET.UNIT.TO.BUS" parameter (menu path: TRANSMITTER INFO  PA DATA)
confirm with "Accept" to adapt the scaling of the Analog Input Block automatically to theTransducer Block. The OUT unit is updated accordingly ( Kap. 7.9).

72

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

7.9

Commissioning

System units (SET UNIT TO BUS)

The onsite display of Cerabar S and the MEASURED VALUE (FieldCare) display the samevalue as standard. The bar graph on the onsite display corresponds to the standardized valueof the Analog Input Block. The digital output value of the Analog Input Block OUT works
independently of the MEASURED VALUE or of the onsite display.
The following options are available so that the onsite display or the MEASURED VALUE andthe digital output value display the same value:•Set the values for the lower and upper limit of PV SCALE and OUT SCALE in the Analog
Input Block as equal (see also  ä 72, "Scaling the OUT value"):–LOWER VALUE (PV SCALE) = LOWER VALUE (OUT SCALE)–UPPER VALUE (PV SCALE) = UPPER VALUE (OUT SCALE)•By means of the SET UNIT TO BUS parameter, confirm the option "On". to adapt the scalingof the Analog Input Block automatically to the Transducer Block. The OUT unit assumes
the value of the PV unit.
Example:The onsite display or the MEASURED VALUE and the OUT value display 100 mbar (1.5 psi).
Select the new unit "psi" by means of the PRESS. ENG. UNIT parameter.
•Display–Onsite display and MEASURED VALUE: 1.45 psi (97 psi)
–OUT value: 100 mbar (1.45 psi)
•By means of the SET UNIT TO BUS parameter, confirm the option "On".
Menu path onsite display: GROUP SELECTION OPERATING MENU TRANSMITTERINFO PA DATA
•Result:
The OUT value displays 1.45 psi (97 psi).
In the following instances, the onsite display or the MEASURED VALUE and the digital
output value of the Analog Input Block OUT no longer display the same value:•If you change the operating mode
•If you change the values for PV SCALE•If you change the values for OUT SCALE•If you change the unit of the primary value!

CAUTION

Note Dependencies when setting parameters!
‣If you confirm the SET UNIT TO BUS parameter, please note that a change in the digital
output value could affect the control system

Endress+Hauser

73

Maintenance

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

8Maintenance

Keep the pressure compensation and GORE-TEX® filter (1) free from contamination andwater.

1

P01-PMC71xxx-17-xx-xx-xx-001

8.1

Cleaning instructions

Endress+Hauser offer flushing rings as accessories to clean process isolating diaphragms
without taking the transmitters out of the process.
For further information please contact your local Endress+Hauser Sales Center.

8.1.1

PMP75

We recommend you perform CIP (cleaning in place (hot water)) before SIP (sterilization inplace (steam)) for pipe diaphragm seals.
A frequent use of sterilization in place (SIP) will increase the stress on the process isolating
diaphragm. Under unfavorable circumstances in the long term view we cannot exclude thata frequent temperature change could lead to a material fatigue of the process isolating
diaphragm and possibly to a leakage.

8.2

Exterior cleaning

Please note the following points when cleaning the device:•The cleaning agents used should not attack the surface and the seals.
•Mechanical damage to the process isolating diaphragm, e.g. due to pointed objects, must
be avoided.

74

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Troubleshooting

9

Troubleshooting

9.1

Messages

The following table lists all the possible messages that can occur.
The device differentiates between the error types "Alarm", "Warning" and "Error".
You may specify whether the instrument should react as if for an "Alarm" or "Warning" for"Error" messages. See "Corresponds to NA 64" column and Section 8.2 "Response of outputs
to errors".
In addition, the "Message category NE 107" column classifies the messages in accordancewith NAMUR Recommendation NE 107:
•Failure (F)•Function check (C)•Out of specification (S)•Maintenance required (M)Error message display on the onsite display:•The measured value display shows the message with the highest priority. See "Priority"
column.
•The ALARM STATUS parameter shows all the messages present in descending order ofpriority. You can scroll through all the messages present with the S-key or O-key.
Message display via FieldCare:•The ALARM STATUS parameter shows the message with the highest priority.
See "Priority" column.
The device status (Device functions/Diagnostics menu) displays the status signal, the error
message, cause and remedial measures.
•If the device detects a defect in the onsite display during initialization, special error
messages are generated. For the error messages, see Page 82, Section 8.1.1 "Onsite
display error messages".
•For support and further information, please contact Endress+Hauser Service.
•See also Section 8.4, 8.5 and 8.6.
•The PROFIBUS status is updated depending on the message type, or depending on thesetting for flexible alarms.
Code

Correspond Message
s to NA 64 category
NE 107101 (A101) Alarm
B

Failure (F)Message/description

Cause

F>Sensor electronic
EEPROM error– Wait a few minutes.
– Electromagnetic effects are– Restart the device. Perform reset
greater than specifications in(Code 2506 or 33062).
the technical data.
– Block off electromagnetic effects(See Section 9.) Thisor eliminate source of
message normally only appears
disturbance.
briefly.
– Sensor defect.

MeasurePriority17– Replace sensor.

102(W102)

Warning
C

Maintenanc M>Checksum error in
EEPROM: peakholde required
segment(M)– Main electronics defect. Correct – Replace main electronics.
measurement can continue aslong as you do not need thepeak hold indicator function.

51

106(W106)

WarningC

Functioncheck (C)– Downloading.

50

Endress+Hauser

C>Downloading - please
wait– Wait for download to complete.

75Troubleshooting

Code

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Correspond Message
s to NA 64 category
NE 107110 (A110) Alarm
B

Failure (F)Message/description

Cause

MeasurePriorityF>Checksum error in
EEPROM: configuration
segment– The supply voltage is
disconnected when writing.

6– Reestablish supply voltage.
Perform reset (Code 1 or 40864)
if necessary. Carry out calibration
again.

– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.)– Block off electromagnetic effects
or eliminate sources ofdisturbance.

– Main electronics defect.

– Replace main electronics.

113 (A113) Alarm
B

Failure (F)F>ROM failure intransmitter electronic– Main electronics defect.

– Replace main electronics.

1

115 (E115)

Error
B
Factory
setting:
Warning

Out ofspecification (S)S>Sensor overpressure– Overpressure present.

– Reduce pressure until message
disappears.

29– Sensor defect.

– Replace sensor.

116(W116)

Warning
C

Maintenanc M>Download error,
repeat download
e required(M)– The file is defect.

– Use another file.

120 (E120)

Error
B
Factory
setting:
Warning

Out ofspecification (S)S>Sensor low pressure

121 (A121) Alarm
B

Failure (F)122 (A122) Alarm
B

Failure (F)36– During the download, the data – Check cable connection PC –transmitter.
are not correctly transmitted to– Block off electromagnetic effects
the processor, e.g. because ofor eliminate sources ofopen cable connections, spikes
disturbance.
(ripple) on the supply voltage– Perform reset (Code 1 or 40864)
or electromagnetic effects.
and carry out calibration again.
– Repeat download.
– Pressure too low.

– Increase pressure until message
disappears.

– Sensor defect.

– Replace sensor.

F>Checksum error infactory segment of
EEPROM– Main electronics defect.

– Replace main electronics.

F>Sensor not connected– Cable connection sensor –main – Check cable connection and repair 13
electronics disconnected.
if necessary.
– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.)– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Main electronics defect.

– Replace main electronics.

– Sensor defect.

– Replace sensor.

30

5130 (A130) Alarm
B

Failure (F)F>EEPROM is defect.

– Main electronics defect.

– Replace main electronics.

10

131 (A131) Alarm
B

Failure (F)F>Checksum error in
EEPROM: min/max
segment– Main electronics defect.

– Replace main electronics.

9132 (A132) Alarm
B

Failure (F)F>Checksum error in
totalizer EEPROM– Main electronics defect.

– Replace main electronics.

7

133 (A133) Alarm
B

Failure (F)F>Checksum error in
History EEPROM– An error occurred whenwriting.

– Perform reset (Code 1 or 40864)
and carry out calibration again.

8– Main electronics defect.

– Replace main electronics.

– The linearization table is not
monotonic increasing or
decreasing.

– Add to or correct linearizationtable. Then accept linearizationtable again.

602(W602)

76

WarningC

Functioncheck (C)C>Linearization curve
not monoton

55

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Troubleshooting

Code

Correspond Message
s to NA 64 category
NE 107

Message/description

604(W604)

Warning
C

Note! No min. span applies for the Y-points as of software version
58
C>Linearization table"03.10.xx".
not valid. Less than 2points or points too close– The linearization table consists – Add to linearization table. Acceptof less than 2 points.
linearization table again if
necessary.

Functioncheck (C)Cause

MeasurePriority– Correct linearization table and– At least 2 points in theaccept it again.
linearization table are too close
together. A minimum distanceof 0.5 % of the span between
two points must be maintained.
Spans for the "Pressure
linearized" option: HYDR. PRESS
MAX. – HYDR. PRESS MIN.;TANK CONTENT MAX. – TANKCONTENT MIN.
Spans for the "Height linearized"
option: LEVEL MAX – LEVELMIN; TANK CONTENT MAX. –TANK CONTENT MIN.
– Switch off simulation.

58

613(W613)

WarningI

Functioncheck (C)C>Simulation is active– Simulation is switched on, i.e.
the device is not measuring atpresent.

616(W616)

WarningI

Functioncheck (C)C>Simulation is active(AI)

58– Switch off simulation of the AI– Simulation of the AI Block is
Block (ANALOGINPUT BLOCK 
switched on, i.e. the Main
Set the AI STANDARD
Process Value (AI OUT VALUE)
PARAMETER TARGET MODE
that is output does not
to Automatic and set AI
correspond to the sensor signal.
PARAMETER/SIMULATE to No).

700
(W700)

Warning
C

Maintenanc M>Last configuration
not storede required(M)– An error occurred when writing – Perform reset (Code 1 or 40864)
and carry out calibration again.
or reading configuration dataor the power supply was
disconnected.
– Main electronics defect.

702(W702)

WarningC

703 (A703) AlarmB

704 (A704) AlarmB

705 (A705) Alarm
B

Endress+Hauser

Maintenanc M>HistoROM data notconsistent.
e required(M)

Failure (F)Functioncheck (C)

Failure (F)F>Measurement error

C>Measurement error

F>Measurement error

52– Replace main electronics.

– Data were not written correctly – Repeat upload.
– Perform reset (Code 1 or 40864)
to the HistoROM, e.g. if theand carry out calibration again.
HistoROM was detached during
the writing process.

53– HistoROM does not have anydata.

– Copy suitable data to theHistoROM. (See also Page 57,
Section 5.6.1 "Copying
configuration data".)– Fault in the main electronics.

– Briefly disconnect device from the 22
power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 12
power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 21
power supply.

– Main electronics defect.

– Replace main electronics.

77Troubleshooting

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

CodeCorrespond Message
s to NA 64 category
NE 107

706(W706)

WarningC

Message/description

Cause

MeasurePriority

Maintenanc M>Configuration in
HistoROM and device
e required
not identical(M)– Configuration (parameters) in – Copy data from the device to the 57
HistoROM. (See also Page 57,
the HistoROM and in the device
Section 5.6.1 "Copying
is not identical.
configuration data".)– Copy data from the HistoROM tothe device. (See also Page 57,
Section 5.6.1 "Copying
configuration data".) The message
remains if the HistoROM and thedevice have different software
versions. The message goes out if
you copy the data from the device
to the HistoROM.
– Device reset codes such as 1 or
40864 do not have any effect on
the HistoROM. That means that if
you do a reset, the configurationsin the HistoROM and in the device
may not be the same.

707 (A707) Alarm
B

Functioncheck (C)C>X-VAL. of lin. table
out of edit limits.

– At least one X-VALUE in thelinearization table is either
below the value for HYDR.
PRESS MIN. or MIN. LEVEL orabove the value for HYDR.
PRESS. MAX. or LEVEL MAX.

– Carry out calibration again. (See also Operating Instructions
BA00296P or these Operating
Instructions, Page 2.)710(W710)

Functioncheck (C)B>Set span too small.
Not allowed.

– Values for calibration (e.g.
lower range value and upper
range value) are too close
together.

– Adjust calibration to suit sensor. 49(See also Operating
Instructions BA00296P,
parameter description MINIMUM
SPAN or these Operating
Instructions, Page 2.)– The sensor was replaced andthe customer-specificconfiguration does not suit thesensor.

– Adjust calibration to suit sensor.
– Replace sensor with a suitablesensor.

– Unsuitable download carried
out.

– Check configuration and perform
download again.

WarningC

37

713 (A713) Alarm
B

Functioncheck (C)C>100% POINT level out – The sensor was replaced.
of edit limits– Carry out calibration again.

38

715 (E715)

Out ofspecification (S)S>Sensor overtemperature– The temperature measured in
the sensor is greater than theupper nominal temperature of
the sensor. (See also
Operating Instructions
BA00296P, parameter
description Tmax SENSOR orthese Operating Instructions,Page 2.)– Reduce process temperature/
ambient temperature.

32– Unsuitable download carried
out.

– Check configuration and perform
download again.

– Sensor defect.

– Replace sensor.
– Reduce pressure.

716 (E716)

78

Error
C
Factory
setting:
Warning

Error
B
Factory
setting:
Alarm

Failure (F)F>Sensor diaphram
broken24

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

CodeCorrespond Message
s to NA 64 category
NE 107717 (E717)

Error
C
Factory
setting:
Warning

Out ofspecificatio
n (S)

Error
C
Factory
setting:
Warning

Out ofspecification (S)718 (E718)

Troubleshooting

Message/description

Cause

S>Transmitter overtemperature– Reduce ambient temperature.
– The temperature measured in
the electronics is greater thanthe upper nominal temperatureof the electronics (+88 °C).
– Unsuitable download carried
out.

S>Transmitter undertemperature

Measure34– Check configuration and perform
download again.

– Increase ambient temperature.
– The temperature measured in
Insulate device if necessary.
the electronics is smaller thanthe lower nominal temperatureof the electronics (–43 °C).
– Unsuitable download carriedout.

Priority35– Check configuration and perform
download again.

719 (A719) Alarm
B

Functioncheck (C)39
– Carry out calibration again.
C>Y-VAL of lin. table out – At least on Y-VALUE in the( See also Operatingof edit limits
linearization table is below theInstructions BA00296P, chapter
MIN. TANK CONTENT or above
5 or these Operating Instructions,
the MAX. TANK CONTENT.
Page 2.)720 (E720)

Out ofspecification (S)S>Sensor undertemperature

Error
C
Factory
setting:
Warning– Increase process temperature/– The temperature measured inambient temperature.
the sensor is smaller than thelower nominal temperature of
the sensor. ( See also
Operating Instructions
BA00296P, parameter
description Tmin SENSOR or
Operating Instructions, Page 2.)– Unsuitable download carried
out.

– Check configuration and perform
download again.

– Loose connection at sensor
cable– Wait a short period of time andtighten the connection, or avoid
loose connection.

33

721 (A721) Alarm
B

Functioncheck (C)C>ZERO POSITION levelout of edit limits– LEVEL MIN or LEVEL MAX has – Perform reset (Code 35710) andbeen changed.
carry out calibration again.

40

722 (A722) Alarm
B

Functioncheck (C)C>EMPTY CALIB. or
FULL CALIB. out of edit
limits– LEVEL MIN or LEVEL MAX has – Perform reset (Code 35710) andbeen changed.
carry out calibration again.

41

725 (A725) Alarm
B

Failure (F)F>Sensor connection
error, cycle disturbance– Electromagnetic effects are
greater than specifications in
the technical data. (See
Kap. 10.)– Setscrew loose.

– Block off electromagnetic effects
or eliminate source ofdisturbance.
– Retighten setscrew with 1 Nm
(0.74 lbf ft) (see Chap. 4.4.8).

25– Sensor or main electronics
defect.

– Replace sensor or mainelectronics.

– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.)– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Process temperature is outsidepermitted range.

– Check temperature present,
reduce or increase if necessary.

– Sensor defect.

– If the process temperature is
within the permitted range,
replace sensor.

726 (E726)

Error
C
Factory
setting:
WarningEndress+Hauser

Out ofspecification (S)S>Sensor temperatureerror - overrange

31

79Troubleshooting

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

CodeCorrespond Message
s to NA 64 category
NE 107727 (E727)

Error
C
Factory
setting:
Warning

728 (A728) AlarmB

729 (A729) Alarm
B

730 (E730)

Error
C
Factory
setting:
Warning

Out ofspecification (S)Failure (F)Failure (F)Out ofspecification (S)Message/description

Cause

MeasurePriority

S>Sensor pressure error- overrange– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.)– Block off electromagnetic effects
or eliminate source ofdisturbance.

28– Pressure is outside permitted
range.

– Check pressure present, reduce or
increase if necessary.

– Sensor defect.

– If the pressure is within thepermitted range, replace sensor.

– Fault in the main electronics.

– Briefly disconnect device from the 2power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 3
power supply.

– Main electronics defect.

– Replace main electronics.

F>RAM error

F>RAM error

S>LRV user limits
exceeded– Check system/pressure measured 46– Pressure measured value hasvalue.
undershot the value specified
for the Pmin ALARM WINDOW – Change value for Pmin ALARM
WINDOW if necessary. (Seeparameter.
also Operating Instructions
BA00296P, parameter
description Pmin ALARM
WINDOW or these Operating
Instructions, Page 2.)– Loose connection at sensor
cable– Check system/pressure measured 45– Pressure measured value hasvalue.
overshot the value specified for– Change value for Pmax ALARM
the Pmax ALARM WINDOWWINDOW if necessary. (Seeparameter.
also Operating Instructions
BA00296P, parameter
description Pmax ALARM
WINDOW or these Operating
Instructions, Page 2.)731 (E731)

Error
C
Factory
setting:
Warning

Out ofspecification (S)S>URV user limits
exceeded732 (E732)

Error
C
Factory
setting:
Warning

Out ofspecification (S)S>LRV Temp. User limits – Temperature measured value
exceededhas undershot the value
specified for the Tmin ALARM
WINDOW parameter.

– Loose connection at sensor
cable
733 (E733)

80

Error
C
Factory
setting:
Warning

Out ofspecification (S)S>URV Temp. User
limits exceeded– Wait a short period of time andtighten the connection, or avoid
loose connection.

– Check system/temperature
measured value.
– Change value for Tmin ALARM
WINDOW if necessary. (See
also Operating Instructions
BA00296P, parameter
description Tmin ALARM
WINDOW or these Operating
Instructions, Page 2.)48– Wait a short period of time andtighten the connection, or avoid
loose connection.

– Check system/temperature– Temperature measured value
measured value.
has overshot the value specified– Change value for Tmax ALARM
for the Tmax ALARM
WINDOW if necessary. (See
WINDOW parameter.
also Operating Instructions
BA00296P, parameter
description Tmax ALARM
WINDOW or these Operating
Instructions, Page 2.)47

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

CodeCorrespond Message
s to NA 64 category
NE 107736 (A736) Alarm
B

737 (A737) AlarmB

738 (A738) AlarmB

739 (A739) AlarmB

740 (E740)

Error
C
Factory
setting:
Warning

Failure (F)Failure (F)Failure (F)Failure (F)Troubleshooting

Message/description

Cause

MeasureF>RAM error– Fault in the main electronics.

– Briefly disconnect device from the 4
power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 20
power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 19
power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 23
power supply.

– Main electronics defect.

– Replace main electronics.

F>Measurement error

F>Measurement error

F>Measurement error

Maintenanc M>Calculation overflow, – Level measuring mode: Level
mode* "LINd. MEASURAND.":
bad configuration,e required
the measured pressure has
hardware defect(M)
undershot the value for HYDR.
PRESS. MIN. or overshot thevalue for HYDR. PRESS MAX.
(*For other level modes: The
measured level did not reachthe LEVEL MIN value or
exceeded the LEVEL MAX
value.)– Pressure measuring mode:
Main electronics defect.

Priority– Check configuration and carry out 27
calibration again if necessary.
– Select a device with a suitablemeasuring range.
– See also Operating Instructions
BA00296P, parameter
description LEVEL MIN. these
Operating Instructions, Page 2.

– Replace main electronics.

741 (A741) Alarm
B

Functioncheck (C)C>TANK HEIGHT out ofedit limits– LEVEL MIN or LEVEL MAX has – Perform reset (Code 35710) andbeen changed.
carry out calibration again.

43

742 (A742) Alarm
B

Failure (F)F>Sensor connection
error (upload)– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.) This message
normally only appears briefly.

– Wait a few minutes.
– Perform reset (Code 35710) andcarry out calibration again.

18– Cable connection sensor –main – Check cable connection and repair
electronics disconnected.
if necessary.

743 (A743) AlarmB

744 (A744) Alarm
B

745(W745)

Warning
C

Endress+Hauser

Failure (F)Failure (F)F>Electronic PCB error
during initialization

F>Main electronic PCB
error

Maintenanc M>Sensor data
unknowne required(M)– Sensor defect.

– Replace sensor.

– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.) This message
normally only appears briefly.

– Wait a few minutes.
– Restart the device. Perform reset(Code 2506 or 33062).

– Main electronics defect.

– Replace main electronics.

– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.)– Restart the device. Perform reset(Code 2506 or 33062).
– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Main electronics defect.

– Replace main electronics.

– Sensor does not suit the device(electronic sensor nameplate).
Device continues measuring.

– Replace sensor with a suitablesensor.

14

11

54

81

Troubleshooting

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

CodeCorrespond Message
s to NA 64 category
NE 107

Message/description

Cause

MeasurePriority746(W746)

WarningC

C>Sensor connection
error - initializing– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.) This message
normally only appears briefly.

– Wait a few minutes.
– Restart the device. Perform reset(Code 1 or 40864).
– Block off electromagnetic effects
or eliminate source ofdisturbance.

26– Overpressure or low pressure
present.

– Reduce or increase pressure.

Functioncheck (C)

747 (A747) Alarm
B

Failure (F)F>Sensor software not– Sensor does not suit the devicecompatible to electronics(electronic sensor nameplate).

– Replace sensor with a suitablesensor.

16

748 (A748) Alarm
B

Failure (F)F>Memory failure insignal processor– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 9.)– Block off electromagnetic effects
or eliminate source ofdisturbance.

15– Main electronics defect.

– Replace main electronics.

– By means of the operation
profile, options were selected
for the configuration of thedevice but the options do notsuit one another. For example,
if the option "1" (linearizationtable) was selected for
LIN_TYPE and the unit "1347(m3/s)" was selected for
PRIMARY_VALUE_UNIT.

– Check configuration.
– Perform reset (Code 1 or 40864)
and carry out calibration again.

750 (A750) Warning
C

Functioncheck (C)C>Configuration not
permitted

9.1.1

44

Onsite display error messages

If the device detects a defect in the onsite display during initialization, the following error
messages can be displayed:
Message

MeasureInitialization, VU Electr. Defect A110

Exchange onsite display.

Initialization, VU Electr. Defect A114
Initialization, VU Electr. Defect A281
Initialization, VU Checksum Err. A110
Initialization, VU Checksum Err. A112
Initialization, VU Checksum Err. A171

82

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

9.2

Troubleshooting

Response of outputs to errors

The device differentiates between the message types "Alarm", "Warning" and "Error".
 See the following table and Page 75, Section 8.1 "Messages".
OutputA (Alarm)

W (Warning)

PROFIBUS

The process variable in question istransmitted with the status Bad.

Device continues measuring. The process For this error, you can enter whether thevariable in question is transmitted with device should react as in the event of analarm or as in the event of a warning.
the status Uncertain.
The output status is transmitted
accordingly with the status Bad,
Uncertain, or GOOD. To configure thestatus for this error, you can configurethe "SELECT ALARM TYPE" parameter(see BA00296P) or the relevant
parameter in Fieldcare (menu path:
PROFILE VIEW  PHYSICAL BLOCK PB PARAMETER  PV STATUS CONFIG( Kap. 9.2.2)). Note: "GOOD" can only
be configured as the status to be output
via Fieldcare in the "PV STATUS CONFIG"
menu path.

Bar Graph(onsite display)

The bar graph assumes the values
Device continues measuring.
specified via the FAILSAFE MODE (FAIL
SAFE MODE1)) and the FAIL SAFE
DEFAULT VALUE (FAIL SAFE DEFAULTVALUE1) parameters.
 See also Section 8.2.1.

For this error, you can enter whether thedevice should react as in the event of analarm or as in the event of a warning.
See corresponding "Alarm" or "Warning"column.

Onsite display– The measured value and message are – The measured value and message aredisplayed alternately
displayed alternately-symbol– Measured value display:-symbol is – Measured value display:
flashes.
permanently displayed.

– The measured value and message aredisplayed alternately– Measured value display: seecorresponding "Alarm" or "Warning"columnMessage display:
Message display– A + 3-digit number such as A122 and – W + 3-digit number such as W613
and– Description– Description

Message display:– E + 3-digit number such as E713 and– Description

In the case of an alarm, the ALARM
STATUS parameter displays a 2) 3-digit
number such as 122 for "Sensor
connection error, incorrect data."

In the case of an error, the ALARM
STATUS2 parameter displays a 3-digit
number such as 731 for "Pmax ALARM
WINDOW undershot".

Remote operation(FieldCare)In the case of a warning, the ALARM
STATUS2 parameter displays a 3-digit
number such as 613 for "Simulation is
active".

E (Error: Alarm/Warning)

1) Parameter are displayed via remote operation (e.g. FieldCare) only.
Menu path: PROFILE VIEW  ANALOG INPUT BLOCK  AI PARAMETER  FAIL SAFE MODE
2) Menu path onsite display: GROUP SELECTION  OPERATING MENU  DIAGNOSTICS MESSAGES
Menu path FieldCare: MANUFACTOR VIEW  OPERATING MENU  DIAGNOSTICS MESSAGESEndress+Hauser

83Troubleshooting

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

9.2.1

Analog Input Block

If the Analog Input Block receives an input or simulation value with the status Bad, theAnalog Input Block uses the failsafe mode defined in the FAIL SAFE MODE1 parameter.
The following options are available by means of the FAIL SAFE MODE1 parameter:•Last valid OutValue
The last valid value is used for further processing with the status Uncertain.
•FAIL SAFE DEFAULT VALUE
The value specified by means of the FAIL SAFE DEFAULT VALUE1 parameter is used forfurther processing with the status Uncertain.
•Status bad
The current value is used for further processing with the status Bad.
Factory setting:•FAIL SAFE MODE1: FAIL SAFE DEFAULT VALUE•FAIL SAFE DEFAULT VALUE1: 0•The failsafe mode is also activated if the "Out of Service O/S" option was selected by meansof the TARGET_MODE2 parameter.
•The FAIL SAFE MODE and FSAVE_VALUE parameter are available via remote operation(e.g. FieldCare) only.
1) Menu path: PROFILE VIEW ANALOG INPUT BLOCK AI PARAMETER
2) Menu path: PROFILE VIEW ANALOG INPUT BLOCK AI STANDARD PARAMETER

9.2.2

Setting the status of the flexible alarms

The event category can be defined individually for the following events - regardless of theevent group to which they are assigned in the default setting:•115: Sensor overpressure•120: Sensor low pressure•715: Sensor over temperature•716: Process isolating diaphragm broken•717: Transmitter over temperature•718: Transmitter under temperature•720: Sensor under temperature•726: Sensor temperature error - overrange•727: Sensor pressure error - overrange•730: LRV user limits exceeded•731: URV user limits exceeded•732: LRV Temp. User limits exceeded•733: URV Temp. User limits exceeded•740: Calculation overflow, bad configuration
To change the measured value status (Bad, Uncertain, Good) assigned to an event, select thedesired status from the picklist.
ExampleThe status "Bad" should be used for error 115 "Sensor overpressure" instead of the status"Uncertain".

84

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Troubleshooting

1.In the FieldCare navigation window navigate to PROFILE VIEW PB Parameter

2.In the default setting, all the bits have "Uncertain" for "Status Select Events", apart from 716.

3.Select the "Bad" option for the row "Status Select Event 115". Press the Enter key to confirm.

9.3

Confirming messages

Depending on the settings for the ALARM DISPL. TIME and ACK. ALARM MODE
parameters, the following measures should be taken to clear a message:
Settings 1)

Measures– ALARM DISPL. TIME = 0 s– ACK. ALARM MODE = Off– Rectify cause of the message (see also Section 8.1).

– ALARM DISPL. TIME > 0 s– ACK. ALARM MODE = Off– Rectify cause of the message (see also Section 8.1).
– Wait for the alarm display time to elapse.

– ALARM DISPL. TIME = 0 s– ACK. ALARM MODE = On

– Rectify cause of the message (see also Section 8.1).
– Confirm message using ACK. ALARM parameter.

– ALARM DISPL. TIME > 0 s– ACK. ALARM MODE = On

– Rectify cause of the message (see also Section 8.1).
– Confirm message using ACK. ALARM parameter.
– Wait for the alarm display time to elapse. If a message appears and thealarm display time elapses before the message has been acknowledged,
the message will be cleared once it has been acknowledged.

1) ALARM DISPL. TIME and ACK. ALARM MODE parameter are located in the MESSAGES menu.

Endress+Hauser

85Troubleshooting

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

9.4

Repair

The Endress+Hauser repairs concept provides for measuring devices to have a modular
design and also the customer may carry out repairs ( ä 86, "Spare Parts).
•For certified devices, please consult Chapter "Repair of Ex-certified devices".
•For more information on service and spare parts contact the Endress+Hauser Service.
( See www.endress.com/worldwide)

9.5!

Repair of Ex-certified devices

WARNING

Limitation of electrical safety due to incorrect connection!
Risk of explosion!
When repairing Ex-certified devices, please note the following:•Only specialist personnel or Endress+Hauser may undertake repairs of certified devices.
•Relevant standards, national hazardous area regulations and Safety Instructions (XA...)and Certificates must be observed.
•Only genuine Endress+Hauser spare parts may be used.
•When ordering spare parts, please check the device designation on the nameplate.
Identical parts may only be used as replacements.
•Electronic inserts or sensors already in use in a standard instrument may not be used asspare parts for a certified device.
•Carry out repairs according to the instructions. After repairs, the device must fulfil therequirements of the specified individual tests.
•A certified device may only be converted into another certified variant by Endress+Hauser.
•All repairs and modifications must be documented.

9.6

Spare Parts•Some replaceable measuring device components are identified by means of a spare part
nameplate. This contains information about the spare part.
•All the spare parts for the measuring device along with the order code are listed In the
W@M Device Viewer (www.endress.com/deviceviewer) and can be ordered. If available,
users can also download the associated Installation Instructions.
Measuring device serial number:•Located on the device and spare part nameplate.
•Can be read out via the "DEVICE SERIAL No" parameter in the "TRANSMITTER DATA"
submenu.

9.7

ReturnThe measuring device must be returned if repairs or a factory calibration are required, or if
the wrong measuring device has been ordered or delivered. According to legal regulations,
Endress+Hauser, as a ISO-certified company, is required to follow certain procedures whenhandling returned products that are in contact with process fluids.
To ensure swift, safe and professional device returns, please read the return procedures andconditions on the Endress+Hauser website at www.services.endress.com/return-material.

9.8

Disposal

When disposing, separate and recycle the device components based on the materials.

86

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

9.9
Date

12.2004

Software version

03.00.zz

Troubleshooting

Software history

Changes software

Documentation
Operating Instructions

Description ofInstrument Functions

BA295P/00/EN/11.04
52025890—

BA295P/00/EN/11.05
71009595

BA296P/00/EN/01.05
52026474– Operation via onsite display with three keys
implemented.
– New level modes "Level Easy Pressure" and "Level Easy
Height" implemented.
– DOWNLOAD FUNCTION parameter added to
OPERATION group.
– Factory settings for the "Error"-type messages
redefined.
– Chinese" and "Japanese" menu language included as
standard.

BA295P/00/EN/10.07
71043300

BA296P/00/EN/07.07
71043302

BA295P/00/EN/12.07
71043300

BA296P/00/EN/07.07
71043302

BA295P/00/EN/05.08
71071780

BA296P/00/EN/05.08
71071861

BA295P/00/EN/08.08
71077546

BA296P/00/EN/05.08
71071861

Compatible with:– FieldCare from Version 2.15.00

BA295P/00/EN/06.09
71095436

BA296P/00/EN/06.09
71095454

BA295P/00/EN/05.10
71095436

BA296P/00/EN/05.10
71111811

Original software.
Compatible with:– Update ToF Tool – Field Tool® Package, version 2.03
or higher

05.200704.00.zz

BA00295P/00/EN/13.11 BA00296P/00/EN/13.11
71139781
71139803
BA00295P/00/EN/14.12 BA00296P/00/EN/13.11
71161900
71139803
07.201304.01.zz

Profile 3.02 Integration

BA00295P/00/EN/15.13 BA00296P/00/EN/14.13
71220562
71221509
BA00295P/00/EN/16.14 BA00296P/00/EN/14.13
71254308
71221509
BA00295P/00/EN/17.14 BA00296P/00/EN/15.14
71270363
71270404
BA00295P/00/EN/18.16 BA00296P/00/EN/18.16
71335661
71336361

9.10

Endress+Hauser

Hardware history

Date

Hardware version

Changes hardware

05.2005

1.0

Original hardware

06.2007

1.10

Resistance inserted due to new requirements.

04.2008

02.00

Replacement of IC Media Access Unit

87

Technical data

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

10Technical data

For technical data, please refer to the Technical Information TI00383P for Cerabar S.

88

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Index
A

N

C

O

Acyclic data exchange . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Alarm messages. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Cable specification . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
Cerabar S block model. . . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Configuring the device address . . . . . . . . . . . . . . . . . . . . 36
Current consumption . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
Cyclic data diagram . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 42
Cyclic data exchange . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40

D

Data format . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 52
Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Diaphragm seals, installation instructions . . . . . . . . . . . 17
Diaphragm seals, vacuum application. . . . . . . . . . . . . . . 18
Display. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29

EElectrical connection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
Error messages. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75

F

Factory setting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 60
FieldCare. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56

G

General structure of the operating menu . . . . . . . . . . . . 53
Grounding. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
GSD (Device Master File) . . . . . . . . . . . . . . . . . . . . . . . . . 38

H

Hazardous area . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Heat insulation. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20
HistoROM/M-DAT . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56

I

Incoming acceptance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Input data, Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 42
Installation instructions for devices with diaphragm seals
17
Installation instructions for devices without diaphragmseals. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

L

Language, selection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 63
Level measurement . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 68
Level measurement, Quick Setup menu . . . . . . . . . . . . . 70
Locking operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59

M

Measuring arrangement for level measurement. . . . . . 17
Measuring arrangement for pressure measurement . 15–
16
Measuring mode, selection . . . . . . . . . . . . . . . . . . . . . . . 63

Endress+Hauser

Nameplate . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Number of devices. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33
Onsite display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Operating elements, function . . . . . . . . . . . . . . . . . . 31–32
Operating elements, position. . . . . . . . . . . . . . . . . . . . . . . 30
Operating keys, onsite, function . . . . . . . . . . . . . . . . . . . . 32
Operating keys, position. . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Operational safety. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Output data, Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . 42
Overvoltage protection . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28

P

Pipe mounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
Position adjustment, onsite . . . . . . . . . . . . . . . . . . . . . . . . 31
Position adjustment, onsite, FieldCare . . . . . . . . . . . . . . . 65
Pressure measurement. . . . . . . . . . . . . . . . . . . . . . . . . . . . 66
Pressure measurement, Quick Setup menu . . . . . . . . . . . 66
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

Q

Quick Setup menu level . . . . . . . . . . . . . . . . . . . . . . . . . . . 70
Quick Setup menu pressure . . . . . . . . . . . . . . . . . . . . . . . . 66

R

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Repair of Ex-certified devices. . . . . . . . . . . . . . . . . . . . . . . 86
Reset . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 60
Returning devices . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Rotating the housing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

S

Scaling the OUT value. . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Scope of delivery . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12
Separate housing, assemble and mount. . . . . . . . . . . . . . 22
Shielding. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Slot/index tables . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 46
Software history . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 87
Spare Parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Status code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43
Storage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Supply voltage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27
System architecture PROFIBUS PA . . . . . . . . . . . . . . . . . . 33
System integration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 38
System units (SET UNIT TO BUS) . . . . . . . . . . . . . . . . . . . 73

T

Temperature isolator, installation instructions. . . . . . . . 19
Troubleshooting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75

U

Unlocking operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59

89

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

W

Wall mounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
Warnings . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Welding recommendation. . . . . . . . . . . . . . . . . . . . . . . . . 23
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

90Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with PROFIBUS PA

Endress+Hauser

91

71335661

www.addresses.endress.com