https://kamemizu.co.jp/e_kamemizu

219AFBZX00118000
Dynamic impression lining material

DIL
INSTRUCTIONS FOR USE
Ⅰ．INTRODUCTION
DIL is a resin for hard lining material ofdentures base.And DIL begins to starthardening after taking the functional
impression for a week.
Ⅱ．INDICATIONS
It is used for denture lining.
Ⅲ．CONTRAINDICATIOＮ
It should not be used with those patients
who had allergic reactions including rash,skin inﬂammation to methyl methacrylate
polymer, methyl methacrylate monomer ororganic solvents in the past.
Ⅳ．POSSIBLE SIDE EFFECT
While it is used, allergic reactions including
redness, sore, blister, swelling may occur.
Ⅴ．INCOMPATIBILITIESDo not use metal plates.
Ⅵ．PRECAUTIONS

１．Safety precautions
1. If allergic reactions including rash, eczema,
redness, tumor, swelling, itch, eruption,
numbness occur to a patient due to using
the material, stop using it immediately and let
the patient call a doctor, or call a doctor incase of a dentist or a medical staﬀ suﬀered.
2. When using the material, take caresuﬃciently so as not to contact with eyes,
skin, or clothing by wearing medical gloves
and safety glasses. Although medical gloves(for dentistry) prevent it from contactingwith hands directly, as it is known, organic
solvents or monomers penetrate them in ashort time. So, if the liquid, the powder, themixture, or the bonding material sticks togloves, throw them away immediately, andrinse hands thoroughly with running waterwith soap.
3. Take suﬃcient precautions so that thematerial will not contact with patient seyes, skin, or clothing. In addition, try not to
let it stick to any other parts of mucous inthe mouth as much as possible than thepart that it should be applied.
4. If it contacts with patient s eyes, rinsethem with plenty of running water
immediately, and let him call an eye
doctor, or call an eye doctor in case of adentist or a medical staﬀ suﬀered.
5. If excess resin ﬂows into the respiratory
tract or the esophagus, let a patient calla medical specialist and have a treatment.
6．After having ﬁnished treatment in themouth, immediately clean the mouth
suﬃciently (let a patient rinse the mouththoroughly). And if the material sticks tothe skin, rinse with plenty of runningwater with soap. If it sticks to clothing,
immediately wipe it oﬀ with alcohol (withcotton ball) thoroughly.

２. Handling and manipulation precautions

1. When a mucous membrane becomesinﬂamed or there is a big diﬀerence in theupper jaw of a full denture, quantity ofpowder and liquid is twice as much ofthat mentioned above.
3. Put the liquid in a mixing cup ﬁrst andadd the powder, then mix immediately
for about 20 seconds.
4. Spread the mixture onto the inner surfaceof the denture base, so that the surface
becomes even as much as possible byusing a spatula. At the room temperatureof 23℃、it is possible to heap it after 1 or
2 minutes from the beginning of mixing.
5. When the viscosity of the mixture becomesappropriate, insert the denture in the mouthfor occlusion. After 4 to 5 minutes after
insertion, the denture can be taken out.
6. After taking out the denture, cut any
excess of the mixture with a design knife
or scissors.
7. Apply New top coat as a surface
smoothing material to the fully dried lining
surface, and the part where mucous
surface meets periphery thoroughly.
After that, leave the denture to dry(approx.15minutes), or dry it using an air
syringe. Then, rinse it well with runningwater, and when it does not smell any
more, let the patient wear it.
8. Depending on the temperature in themouth, polymerization begins voluntarily
one week after wearing the denture,and it will be gradually polymerized. If it
is not inserted in the mouth longer, thehardening time will be required longer.
9. For the ﬁrst week or so after lining, therelined denture should be rinsed with cold
water only, and after that Polident withenzyme (Earth pharmaceutical company)
can be used. Other denture cleaners on
the market should not be used, as theycause rough surface.

distribution of bite pressure against themucous beneath the denture base, themucous membrane needs treatment
before hand.
2. As the liquid is inﬂammable, it should
not be used or placed near the ﬁre. And
if it is spilt on the table or the ﬂoor,
wipe it immediately with dry cloth.
3. It should be used in the place where theventilation is appropriate (severalventilation per hour).
4. When polymerized denture is ground orpolished, use a local ventilation or dust
mask approved by public organizations
not to inhale ﬁne particles.
5. In order to prevent infection, syringes
and spatulas should be cleaned, andsterilized with alcohol after use.
6. It should not be used together withother products.
7. It should not be used in any ways otherthan resin for denture base repair.
8. It should not be used by any person
other than qualiﬁed dentists.

３．Storage precautions

1. The material should be stored to avoid
ﬁre, high temperature, high humidityand direct sunlight, at 0℃ to 23℃.
2.The ingredient of liquid material may
separate into two layers at very lowtemperature(0゜C = 32゜Ｆ or less).
In such a case, please stir strongly
after warming to room temperature.
By using only supernatant ﬂuid, it does
not function properly.
3. It should not be stored in a large quantity
in the same storage. A ﬁre extinguishershould be installed in the place where it isused and the storage.
4. As for both the powder and the liquid, lidsof the containers should be tightened ﬁrmly.
5. While it is not used for a long period oftime, or stored laying them on their side,store it with both a lid and an innerstopper tightened. (Do not throw away theinner stopper and keep it for storage.)
6. The material should be properly storedand controlled, and not be touched byany person other than people engage indentistry

Ⅶ．COMPONENTS
１．Powder 50g
Poly(ethyl methacrylate), coloring agent,and others
２．Liquid45mL
Polyfunctional methacrylate, and others
３．New top coat 30mL
Ethyl acetate , and others
４．Accessories
Powder measuring cup, Mixing cup,Pipette, Spatula, Steeple-crowned cap.
Ⅷ．CLINICAL PROCEDURES
1. Remove an old resin surface andperipheral edge of a denture base toobtain a new surface, and rinse withwater and dry it. Then apply a bonding
material New top coat to the denturebase.
2. Measure both powder and liquid. Incase of lower jaw of a full denture, thestandard ratio of powder to liquid is thepowder adjusting the scale of 3 (2.8g)of measuring cup to liquid adjusting thescale 2.5 (2.5mL) of Pipette. In case of[CAUTION]
The mixture should not be spread toomuch, as it will cause to ﬂow into thepharynx. Ask the patient to breath
through his nose so that it will not ﬂow
into the pharynx, which should be
conﬁrmed in the case of patients who has
lower reﬂex (Such as aged people).
[WARRANTY]
KAMEMIZU CHEMICAL IND.CO., LTD. will
replace any product that is proved to be
defective. KAMEMIZU CHEMICAL IND.CO.,
LTD. does not accept liability for any loss
or damage, direct, consequential orspecial, arising out of the application or
use of or the inability to use theseproducts. Before using, the user shalldetermine the suitability of the products
for the intended use and the user
assumes all risk and liability whatsoeverin connection therewith.

KAMEMIZU CHEMICAL IND.CO.,LTD.
17-16 TOYOSATO-CHO, NEYAGAWA,OSAKA 572-0071, JAPANEC

REP

Emergo Europe
Molenstraat 152513 BH The Hague
The Netherlands
Tel : (31) (0) 70 345-8570
Fax: (31) (0) 70 346-7299