ENGLISH – USER INSTRUCTIONS

4. Attach the #7920 Piggyback Adapter onto the #7006 Probed Cartridge.

CONTENTS:
2 #7006 Probed Cartridges (one for #7000 and one for #9000)
1 Pair #7940P100 Filter Disks
1 Pair #7920 Filter Disk Piggyback Adapters

5. Attach a new #7940P100 Filter Disk onto the #7920 Piggyback Adapter, making sure that the #7920
gasket is not dirty or damaged and is in place. If it is not, then replace with a new one and do not attachwithout a gasket. (see figure E for #7000, Figure F for #9000)

INSTRUCTIONS:
Before conducting a quantitative fit test, the #7000 or #9000 Series Respirator Facepiece must be
assembled with the appropriate #7006 probed cartridge and the #7940P100 Filter Disk together with a#7920 Piggyback Adapter on one side, and a #7940P100 Filter Disk on the other side.

7000/9000 SERIES RESPIRATORS#7006 PROBED CARTRIDGE KIT INSTRUCTIONSFOR QUANTITATIVE FIT TESTING WITH
PORTACOUNT®

#7000 – Figure E#9000 – Figure F

Assembly is as follows:
Before attaching the appropriate #7006 probed cartridge, inspect the facepiece to cartridge sealingsurface. Make sure it is clean and undamaged.
1. Insert tube from cartridge into the cartridge retainer opening in the facepiece that is closest to the dot(see figure A).
Note: when used with the #9000 full face respirator you must also insert the longer tube into the inhale
opening in the nosecup, and past the rubber diaphragm (see figure B).
#7000 – Figure A#9000 – Figure B

6. Make a careful inspection of the #7920 and #7940P100 to ensure they are fully sealed.
7. Attach a #7940P100 onto the facepiece and check to see it is properly sealed. (see figure Gfor #7000, Figure H for #9000) Check the inhalation diaphragm for dirt or damage and see that itis properly seated.
#7000 – Figure G#9000 – Figure H

WARNING
The #7006 Probed Cartridge is for use in quantitative fit testing together with a TSI Portacount or equivalent fit
test instrument. The cartridge does not contain filter media and therefore provides no protection.
In accordance with OSHA 29 CFR 1910.134 (f) a qualitative or quantitative fit test must be performed before arespirator is assigned to a wearer. The Moldex #7000 half mask and/or #9000 full face respirators can bequantitatively fit tested using the TSI Portacount, or similar, fit tester together with the appropriate Moldex#7006 Probed Cartridge as labeled.
Caution: Always place the #7006 Probed Cartridge on the user’s left side of respirator. There are two different
probed cartridges per kit. ONLY use the shorter tubed cartridge with the #7000. ONLY use the longer tubed
cartridge for the #9000 Read cartridge labels for the correct probed cartridge for either the #7000 or the#9000. Failure to follow these directions will result in incorrect fit test readings.

2. Align the triangle icons on both the facepiece and #7006 cartridge, then rotate the cartridge clockwise toinstall (see figure C for #7000, Figure D for #9000).
Caution: Make sure that the tube is not kinked or obstructed in any way.
#7000 – Figure C#9000 – Figure D
8. Attach the tube to the Quantitative Fit Tester and follow the instructions from the Fit Tester Manufacturer.

CANADIAN CUSTOMER SERVICE
Tel: +1 (800) 421-0668, Ext. 517 Fax: +1 (310) 837-9563®

MOLDEX-METRIC, INC.
10111 W. Jefferson Blvd., Culver City, CA 90232
Tel: +1 (800) 421-0668 or +1 (310) 837-6500
Fax: +1 (310) 837-9563
Email: sales@moldex.com www.moldex.com
Moldex, Ideas that wear well and the PVC-Free logo are
registered trademarks. U.S. Patents Pending.
7006-022 REV A 11/10

LIMITED WARRANTY IMPORTANT NOTICE TO PURCHASERBRAZIL – MOLDEX-METRIC LTDA
Av Ipiranga, 104, 22º Andar Sala 01
Sao Paulo, SP Brazil
Tel/Fax: +55 (11) 26 59 21 08
moldexbrazil@moldex.com

This limited warranty is made in lieu of the warranties of merchantability, fitness for particularpurposes and all other warranties, express or implied. There are no other warranties which extend
beyond the description on the face hereof. The physical standards and specifications of Moldex will be met
by products sold. Exclusive Remedies: damages for the breach of this limited warranty are limited to thereplacement of such quantity of Moldex products proved to be defectively manufactured. Except as provided above, Moldex shall not be liable or responsible for any loss, damage, or liability, direct, indirect, incidental, special, or consequential, arising out of sale, use, or misuse, or the inability to use products by the user.

AUSTRALIA – MOLDEX-METRIC, INC.
Unit 30/14-16 Stanton Road, Seven Hills, NSW 2147
Tel: +61 (2) 9838 7588 Fax: +61 (2) 9838 7544
Email: moldex@bigpond.net.au www.moldex.com
NEW ZEALAND – MOLDEX-METRIC, INC.
3 Spence Road, Henderson, Auckland, NZ
Tel: +64 (9) 837 8247 Fax: +64 (9) 837 8248
Email: trevorc@moldex.com www.moldex.com
MOLDEX TECHNICAL SERVICE DEPARTMENT:+1 (800) 421-0668 or +1 (310) 837-6500, ext. 512/550

3. Check to see that the probed #7006 cartridge is seated properly, both inside and outside the facepiece.
Check the inhalation diaphragms for dirt or damage and check that they are seated properly.

ESPAÑOL – INSTRUCCIONES DEL USUARIO

4. Junte el Adaptador #7920 al Cartucho Sondeado #7006.

CONTENIDO:
2 Cartuchos Sondeados #7006 (uno para #7000 y otro para #9000)
1 Par Discos Filtro #7940P100
1 Par Discos Filtro Adaptador #7920
INSTRUCCIONES:
Antes de llevar a cabo una prueba de ajuste cuantitativa, la máscara del Respirador Series #7000 o #9000
debe ser ensamblada con el cartucho sondeado #7006 apropiado y el Disco Filtro #7940P100 junto con
un Adaptador #7920 en un lado y un Disco Filtro #7940P100 en el otro lado.

SERIES DE RESPIRADORES
7000/9000

KIT DE INSTRUCCIONES DEL CARTUCHOSONDEADO #7006
PARA PRUEBAS DE AJUSTE CUANTITATIVAS
CON PORTACOUNT®

5. Junte un Disco Filtro #7940P100 nuevo al Adaptador #7920, asegúrese que el empaque del #7920 noesté sucio o dañado y que esté en su lugar. Si no lo está, entonces reemplácelo con uno nuevo y no lo
junte sin un empaque. (ver figura E, figura F – muestra #7000 y #9000 ensamblados en un lado y sin
nada en el otro)#7000 – Figura E#9000 – Figura F

Indicaciones de Ensamblado:
Antes de juntar el cartucho sondeado #7006 apropiado, inspeccione la superficie de sellado de la máscara.
Asegúrese que esté limpia y sin daño.
1. Inserte el tubo del cartucho en la abertura del retenedor del cartucho en la máscara que esté más cercana al punto (ver figura A – muestra 7000). Nota: Cuando se usa con el respirador de cara completa#9000 usted también debe insertar el tubo más largo en la abertura para inhalar de la copa nasal ypasar la liga por el diafragma (ver figura B – muestra 9000).
#7000 – Figura A#9000 – Figura B

6. Haga una inspección cuidadosa del #7920 y el #7940P100 para asegurarse que estén completamentesellados.
7. Junte el #7940P100 a la pieza facial y revisé que esté apropiadamente sellada. (ver figura G, figura H –muestra completa el 7000 y 9000 con todas las partes ensambladas). Revisé el diafragma de inhalación
por cualquier suciedad o daño y vea que esté apropiadamente asentado.
#7000 – Figura G#9000 – Figura H

AVISO:El Cartucho Sondeado #7006 es para usarse en pruebas de ajuste cuantitativas junto con un Portacount TSI o
un instrumento equivalente de pruebas de ajuste. El cartucho no contiene un medio de filtrado y por lo tanto noprovee protección.
De acuerdo con OSHA 29 CFR 1910.134 (f) una prueba de ajuste cuantitativa o cualitativa debe ser ejecutada
antes de que sea asignado un respirador a un usuario. La media máscara de Moldex #7000 y/o el respiradorde cara completa #9000 pueden ser usados en pruebas de ajuste cuantitativas usando el Portacount TSI, oalgo similar, si se han usado juntos en pruebas de ajuste con el cartucho sondeado apropiado de Moldex#7000 tal como se ha etiquetado.
Precaución: Siempre coloque el Cartucho Sondeado #7006 en el lado izquierdo del usuario del respirador.
Hay dos cartuchos sondeados diferentes por kit. SOLAMENTE use el cartucho con el tubo más pequeño con el#7000. SOLAMENTE use el cartucho del tubo más largo con el #9000. Lea las etiquetas del cartucho para el
cartucho sondeado correcto ya sea para el #7000 o el #9000. Si falla al seguir estas instrucciones puede
resultar en una lectura incorrecta de las pruebas de ajuste.

2. Alinee los íconos triangulares en la máscara y el cartucho #7006, luego rote el cartucho en sentido delas manecillas del reloj para instalar (vea figura C, figura D – muestra ambos 7000 y 9000).
Precaución: Asegúrese que el tubo no esté obstruído o volteado de alguna forma.
#7000 – Figura C#9000 – Figura D
8. Junte el tubo al Probador de Ajuste Cuantitativo y siga las instrucciones del Fabricante de Probadorde Ajuste.

AVISO IMPORTANTE AL COMPRADOR ACERCA DE LA GARANTÍA LIMITADACANADIAN CUSTOMER SERVICE
Tel: +1 (800) 421-0668, Ext. 517 Fax: +1 (310) 837-9563®

MOLDEX-METRIC, INC.
10111 W. Jefferson Blvd., Culver City, CA 90232
Tel: +1 (800) 421-0668 or +1 (310) 837-6500
Fax: +1 (310) 837-9563
Email: sales@moldex.com www.moldex.com

Moldex e Ideas that wear well son marcas comerciales registradas. Patentes pendientes en EE.UU. El logotipo PVC-Freees una marca comercial de Moldex-Metric, Inc.
7006-022 REV A 11/10

BRAZIL – MOLDEX-METRIC LTDA
Av Ipiranga, 104, 22º Andar Sala 01
Sao Paulo, SP Brazil
Tel/Fax: +55 (11) 26 59 21 08
moldexbrazil@moldex.com
AUSTRALIA – MOLDEX-METRIC, INC.
Unit 30/14-16 Stanton Road, Seven Hills, NSW 2147
Tel: +61 (2) 9838 7588 Fax: +61 (2) 9838 7544
Email: moldex@bigpond.net.au www.moldex.com
NEW ZEALAND – MOLDEX-METRIC, INC.
3 Spence Road, Henderson, Auckland, NZ
Tel: +64 (9) 837 8247 Fax: +64 (9) 837 8248
Email: trevorc@moldex.com www.moldex.com
MOLDEX TECHNICAL SERVICE DEPARTMENT:+1 (800) 421-0668 or +1 (310) 837-6500, ext. 512/550

3. Revisé para ver que el cartucho sondeado #7006 está asentado apropiadamente dentro y fuera de lamáscara. Revisé los diafragmas de inhalación por cualquier suciedad o daño y revisé que estén asentados apropiadamente.

Esta garantía limitada está hecha en lugar de las garantías de comercialización, ajuste a
propósitos particulares y a todas las otras garantías, expresas ó implícitas. No hay otras garantías
que se extiendan más allá de las descripciones en la carátula presente. Los estándares físicos y especificaciones de Moldex serán cumplidos por los productos vendidos. Remedios exclusivos: daños por el
incumplimiento de esta garantía limitada están limitados al reemplazo de esa cantidad de productos Moldex
que se ha probado que han sido fabricados defectuosamente. Excepto por lo provisto arriba, Moldex no es
propenso o responsable por cualquier daño, o tener responsabilidad directa, indirecta, incidental, especial oconsecuencial, proveniente de la venta, uso o abuso, o de la inhabilidad para usar los productos por partedel usuario.