BA01146R/09/EN/08.21
715431282021-07-30

Products

Solutions

Valid as of version
ENU000A, V2.04.xx

Operating Instructions
Ecograph T, RSG35
Universal Data Manager

ServicesEcograph T, RSG35

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 6

1.1
1.2

1.3
1.4

Document function . . . . . . . . . . . . . . . . . . . . . 6
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
1.2.1 Safety symbols . . . . . . . . . . . . . . . . . . 6
1.2.2 Electrical symbols . . . . . . . . . . . . . . . . 6
1.2.3 Symbols forcertain types of information . . . . . . . . . 7
1.2.4 Symbols in graphics . . . . . . . . . . . . . . . 7
Terminology . . . . . . . . . . . . . . . . . . . . . . . . . . 7
Registered trademarks . . . . . . . . . . . . . . . . . . . 8

2

Basic safety instructions . . . . . . . . . . . . 8

2.1
2.2
2.3
2.4
2.5
2.6
2.7

Requirements concerning the staff . . . . . . . . . . 8
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . 8
Operational safety . . . . . . . . . . . . . . . . . . . . . . 9
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . 9
Safety information for desktop version(option) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
IT security . . . . . . . . . . . . . . . . . . . . . . . . . . . . 93

Product description . . . . . . . . . . . . . . . . 10

3.1

Product design . . . . . . . . . . . . . . . . . . . . . . . . 10

4Incoming acceptance and productidentification . . . . . . . . . . . . . . . . . . . . . 10

4.1
4.2
4.3
4.4

Incoming acceptance . . . . . . . . . . . . . . . . . . . 10
Scope of delivery . . . . . . . . . . . . . . . . . . . . . . 10
Product identification . . . . . . . . . . . . . . . . . . . 10
4.3.1 Nameplate . . . . . . . . . . . . . . . . . . . . 10
4.3.2 Name and address of manufacturer . . 11
Storage and transport . . . . . . . . . . . . . . . . . . 11

5

Installation . . . . . . . . . . . . . . . . . . . . . . . 11

5.1
5.2
5.3

Installation conditions . . . . . . . . . . . . . . . . . . 11
5.1.1 Installation dimensions . . . . . . . . . . . 12
Mounting the measuring device . . . . . . . . . . . 12
Post-installation check . . . . . . . . . . . . . . . . . . 13

6

Electrical connection . . . . . . . . . . . . . . 13

6.1
6.2

Connecting requirements . . . . . . . . . . . . . . . . 13
Connecting instructions . . . . . . . . . . . . . . . . . 13
6.2.1 Cable specification . . . . . . . . . . . . . . . 13
Connecting the measuring device . . . . . . . . . . 15
6.3.1 Terminal assignment on the rear ofthe device . . . . . . . . . . . . . . . . . . . . . 15
6.3.2 Supply voltage . . . . . . . . . . . . . . . . . . 15
6.3.3 Relay . . . . . . . . . . . . . . . . . . . . . . . . 15
6.3.4 Digital inputs; auxiliary voltage
output . . . . . . . . . . . . . . . . . . . . . . . . 16
6.3.5 Analog inputs . . . . . . . . . . . . . . . . . . 16

6.3

Endress+Hauser

6.3.6

6.4

Connection example: Auxiliary
voltage output as transmitter power
supply for 2-wire sensors . . . . . . . . . . 17
6.3.7 Connection example: Auxiliary
voltage output as transmitter power
supply for 4-wire sensors . . . . . . . . . . 18
6.3.8 Option: RS232/RS485 interface (rearof device) . . . . . . . . . . . . . . . . . . . . . 18
6.3.9 Ethernet connection (rear of device) . 19
6.3.10 Option: Ethernet Modbus TCP slave . . 20
6.3.11 Option: Modbus RTU slave . . . . . . . . . 20
6.3.12 Connections at front of device . . . . . . 20
Post-connection check . . . . . . . . . . . . . . . . . . 22

7

Operation options . . . . . . . . . . . . . . . . . 23

7.1
7.2

Overview of operation options . . . . . . . . . . . . 23
Structure and function of the operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
7.2.1 Operating menu for operators andmaintenance personnel . . . . . . . . . . . 24
7.2.2 Operating menu for experts . . . . . . . . 25
7.2.3 Submenus and users . . . . . . . . . . . . . 25
Measured value display and operating
elements . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27
Display representation of symbols used inoperation . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
7.4.1 Symbols in operating menus . . . . . . . 29
7.4.2 Symbols in the event logbook . . . . . . . 29
Entering text and numbers (virtual
keyboard) . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Channel color assignment . . . . . . . . . . . . . . . 30
Access to the operating menu via the local
display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Device access via operating tools . . . . . . . . . . 30
7.8.1 Field Data Manager (FDM) analysis
software (SQL database support) . . . . 30
7.8.2 Web server . . . . . . . . . . . . . . . . . . . . 31
7.8.3 OPC server (optional) . . . . . . . . . . . . 31
7.8.4 FieldCare/DeviceCare configuration
software . . . . . . . . . . . . . . . . . . . . . . 31

7.3
7.4

7.5
7.6
7.7
7.8

8

System integration . . . . . . . . . . . . . . . . 32

8.1

Integrating the measuring device in thesystem . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
8.1.1 General notes . . . . . . . . . . . . . . . . . .
8.1.2 Ethernet . . . . . . . . . . . . . . . . . . . . . .
8.1.3 Web server with "Ethernet over USB"
function . . . . . . . . . . . . . . . . . . . . . .
8.1.4 Modbus RTU/TCP slave . . . . . . . . . . .

32
32
32
32
34

9

Commissioning . . . . . . . . . . . . . . . . . . . . 34

9.1
9.2
9.3

Function check . . . . . . . . . . . . . . . . . . . . . . . 34
Switching on the measuring device . . . . . . . . . 34
Setting the operating language . . . . . . . . . . . . 35
3

Table of contents

9.4

9.5
9.6
9.7
9.8
9.9

Configuring the measuring device (Setupmenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
9.4.1 Step-by-step: to the first measured
value . . . . . . . . . . . . . . . . . . . . . . . . 35
9.4.2 Step-by-step: set or delete the limit
values . . . . . . . . . . . . . . . . . . . . . . . . 36
9.4.3 Device setup . . . . . . . . . . . . . . . . . . . 36
9.4.4 Setup via SD card or USB stick . . . . . . 37
9.4.5 Setup via Web server . . . . . . . . . . . . . 37
9.4.6 Setup via FieldCare/DeviceCareconfiguration software . . . . . . . . . . . 38
Advanced settings (Expert menu) . . . . . . . . . . 38
Configuration management . . . . . . . . . . . . . . 39
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Access protection and security concept . . . . . . 39
HTTPS Web server setup . . . . . . . . . . . . . . . . 40

10Operation . . . . . . . . . . . . . . . . . . . . . . . . . 42

10.1

Reading-out and changing the Ethernet
setting address . . . . . . . . . . . . . . . . . . . . . . . 42
10.2 Reading off the device locking status . . . . . . . 42
10.3 Reading off measured values (display
devices) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43
10.4 Web server . . . . . . . . . . . . . . . . . . . . . . . . . . 43
10.4.1 Access to the Web server via HTTP(HTML) . . . . . . . . . . . . . . . . . . . . . . 44
10.4.2 Access to the Web server via XML . . . 44
10.4.3 Setup, operation and service via the
Web server . . . . . . . . . . . . . . . . . . . . 45
10.4.4 Remote control via the Web server . . . 49
10.5 Changing the group . . . . . . . . . . . . . . . . . . . . 49
10.6 Locking operation . . . . . . . . . . . . . . . . . . . . . 49
10.7 Login / logout . . . . . . . . . . . . . . . . . . . . . . . . 50
10.8 SD card/USB stick . . . . . . . . . . . . . . . . . . . . . 50
10.8.1 Function of SD card or USB stick . . . . 50
10.8.2 Functions relating to the SD card or
USB stick . . . . . . . . . . . . . . . . . . . . . 50
10.8.3 Notes on e-mail encryption . . . . . . . . 51
10.8.4 Notes on WebDAV encryption . . . . . . 52
10.8.5 SSL certificates . . . . . . . . . . . . . . . . . 52
10.9 Show data logging . . . . . . . . . . . . . . . . . . . . . 53
10.9.1 Historical data: changing a group . . . . 53
10.9.2 Historical data: Scroll speed . . . . . . . . 53
10.9.3 Historical data: Time scaling . . . . . . . 54
10.9.4 Historical data: Time range
displayed . . . . . . . . . . . . . . . . . . . . . 54
10.9.5 Historical data: Screenshot . . . . . . . . . 54
10.9.6 Historical data: Change the display
mode . . . . . . . . . . . . . . . . . . . . . . . . 54
10.10 Signal analysis . . . . . . . . . . . . . . . . . . . . . . . . 54
10.11 Search in trace . . . . . . . . . . . . . . . . . . . . . . . . 55
10.12 Changing the display mode . . . . . . . . . . . . . . 55
10.13 Adjusting the brightness of the display . . . . . . 55
10.14 Limit values . . . . . . . . . . . . . . . . . . . . . . . . . . 55
10.15 WebDAV client . . . . . . . . . . . . . . . . . . . . . . . 55
10.15.1 Access to the WebDAV server via
HTTP (HTML) . . . . . . . . . . . . . . . . . . 56

4

Ecograph T, RSG35

10.16 Data analysis and visualization with the Field
Data Manager software (FDM) provided . . . . . 56
10.16.1 Structure/layout of a CSV file . . . . . . . 56
10.16.2 Importing UTF-8-encoded CSV files
into spreadsheets . . . . . . . . . . . . . . . 57

11

Diagnostics and troubleshooting . . . 58

11.1
11.2

General troubleshooting . . . . . . . . . . . . . . . . . 58
Troubleshooting . . . . . . . . . . . . . . . . . . . . . . 58
11.2.1 Device error/alarm relay . . . . . . . . . . 58
11.3 Diagnostic information on the local display . . . 59
11.4 Pending, current diagnostic messages . . . . . . . 63
11.5 Diagnosis list . . . . . . . . . . . . . . . . . . . . . . . . . 63
11.6 Event logbook . . . . . . . . . . . . . . . . . . . . . . . . 63
11.7 Device information . . . . . . . . . . . . . . . . . . . . 63
11.8 Diagnostics of measured values . . . . . . . . . . . 63
11.9 Diagnostics of outputs . . . . . . . . . . . . . . . . . . 64
11.10 Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . 64
11.10.1 E-mail test . . . . . . . . . . . . . . . . . . . . 64
11.10.2 Test WebDAV client . . . . . . . . . . . . . 64
11.10.3 Test time synchronization/SNTP . . . . 64
11.10.4 Relay test . . . . . . . . . . . . . . . . . . . . . 64
11.11 Initializing the modem . . . . . . . . . . . . . . . . . . 64
11.12 Resetting the measuring device . . . . . . . . . . . 65
11.13 Clear memory . . . . . . . . . . . . . . . . . . . . . . . . 65
11.14 Reset analysis . . . . . . . . . . . . . . . . . . . . . . . . 65
11.15 Firmware history . . . . . . . . . . . . . . . . . . . . . . 65

12Maintenance . . . . . . . . . . . . . . . . . . . . . . 66

12.1
12.2
12.3

Updating the device software ("firmware") . . . . 66
Instructions for enabling a software option . . . 66
Cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . . 66

13

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . 67

13.1 General notes . . . . . . . . . . . . . . . . . . . . . . . . 67
13.2 Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . 67
13.3 Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 69
13.4 Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . 69
13.4.1 IT security . . . . . . . . . . . . . . . . . . . . . 69
13.4.2 Removing the measuring device . . . . . 69
13.4.3 Disposing of the measuring device . . . 69

14

Accessories . . . . . . . . . . . . . . . . . . . . . . . 70

14.1

Device-specific accessories . . . . . . . . . . . . . . . 70

15Technical data . . . . . . . . . . . . . . . . . . . . 72

15.1 Function and system design . . . . . . . . . . . . . . 72
15.2 Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
15.3 Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
15.4 Power supply . . . . . . . . . . . . . . . . . . . . . . . . 78
15.5 Performance characteristics . . . . . . . . . . . . . . 80
15.6 Installation . . . . . . . . . . . . . . . . . . . . . . . . . . 80
15.7 Environment . . . . . . . . . . . . . . . . . . . . . . . . . 82
15.8 Mechanical construction . . . . . . . . . . . . . . . . 82
15.9 Display and operating elements . . . . . . . . . . . 83
15.10 Certificates and approvals . . . . . . . . . . . . . . . 86
Endress+Hauser

Ecograph T, RSG35

Table of contents

15.11 Ordering information . . . . . . . . . . . . . . . . . . . 86
15.12 Supplementary documentation . . . . . . . . . . . . 87

16

Appendix . . . . . . . . . . . . . . . . . . . . . . . . . 88

16.1

Operating items in the "Expert" menu . . . . . . . 88
16.1.1 "System" submenu . . . . . . . . . . . . . . . 88
16.1.2 "Inputs" submenu . . . . . . . . . . . . . . . 107
16.1.3 "Outputs" submenu . . . . . . . . . . . . . . 129
16.1.4 "Communication" submenu . . . . . . . . 130
16.1.5 "Application" submenu . . . . . . . . . . . 145
16.1.6 "Diagnostics" submenu . . . . . . . . . . . 176

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 180

Endress+Hauser

5About this document

Ecograph T, RSG35

1

About this document

1.1

Document function

These Operating Instructions contain all the information that is required in the various
phases of the device's life cycle: from product identification, incoming acceptance andstorage, to mounting, connection, operation fundamentals and commissioning through totroubleshooting, maintenance and disposal.
Integrated Operating Instructions
At the push of a button, the device displays operating instructions directly on the screen.
This manual complements the operating instructions in the device and explains what is
not directly described in the operating instructions.

1.2

Symbols

1.2.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

1.2.2

Electrical symbols
SymbolMeaning
Direct current
Alternating current
Direct current and alternating current
Ground connection
A grounded terminal which, as far as the operator is concerned, is grounded via agrounding system.
Protective Earth (PE)A terminal which must be connected to ground prior to establishing any otherconnections.
The ground terminals are situated inside and outside the device:• Inner ground terminal: Connects the protectiv earth to the mains supply.
• Outer ground terminal: Connects the device to the plant grounding system.

6

Endress+Hauser

Ecograph T, RSG35

About this document

1.2.3

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Preferred
Procedures, processes or actions that are preferred.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.
Reference to documentation
Reference to pageA

Reference to graphic
Notice or individual step to be observed
1. , 2. , 3. …

Series of steps
Result of a step
Help in the event of a problem
Visual inspection

1.2.4

Symbols in graphics

SymbolMeaning

1, 2, 3,...

Item numbers

,

,

…

Series of steps

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections
Flow direction

A0013441-

A0011187

Hazardous area
Indicates a hazardous area.
Safe area (non-hazardous area)
Indicates a non-hazardous area.

.
A0011188

1.3

Terminology

To improve clarity, abbreviations or synonyms are used in these instructions for thefollowing terms:• Endress+Hauser:
Term used in these instructions: "Manufacturer" or "Supplier"• Ecograph T RSG35:
Term used in these instructions: "Device" or "Measuring device"

Endress+Hauser

7Basic safety instructions

Ecograph T, RSG35

1.4

Registered trademarks

Modbus®
Registered trademark of SCHNEIDER AUTOMATION, INC.
Internet Explorer®, ExcelTM
Registered trademarks of the Microsoft Corporation
Mozilla Firefox®
Registered trademark of Mozilla Foundation
Opera®
Registered trademark of Opera Software ASA.
Google ChromeTM
Registered trademark of Google INC.

2

Basic safety instructions

Reliable and safe operation of the device is guaranteed only if the user reads these
Operating Instructions and complies with the safety instructions they contain.

2.1

Requirements concerning the staff

The personnel for installation, commissioning, diagnostics and maintenance must fulfill
the following requirements:‣ Trained, qualified specialists: must have a relevant qualification for this specificfunction and task
‣ Are authorized by the plant owner/operator‣ Are familiar with federal/national regulations‣ Before beginning work, the specialist staff must have read and understood the
instructions in the Operating Instructions and supplementary documentation as well as
in the certificates (depending on the application)‣ Following instructions and basic conditions
The operating personnel must fulfill the following requirements:‣ Being instructed and authorized according to the requirements of the task by thefacility's owner-operator‣ Following the instructions in these Operating Instructions

2.2

Intended use

This device is intended for the electronic capture, display, recording, analysis, remotetransmission and archiving of analog and digital input signals in non-hazardous areas.
• The manufacturer accepts no liability for damages resulting from incorrect use or use
other than that designated. It is not permitted to convert or modify the device in any
way.
• The device is designed for installation in a panel and must only be operated in an
installed state.

2.3

Workplace safety

For work on and with the device:‣ Wear the required personal protective equipment according to national regulations.
If working on and with the device with wet hands:‣ Due to the increased risk of electric shock, wear suitable gloves.
8

Endress+Hauser

Ecograph T, RSG35

Basic safety instructions

2.4

Operational safety

Risk of injury!

‣ Operate the device only if it is in proper technical condition, free from errors and faults.
‣ The operator is responsible for interference-free operation of the device.
Modifications to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers:‣ If modifications are nevertheless required, consult with the manufacturer.
RepairTo ensure continued operational safety and reliability:‣ Carry out repairs on the device only if they are expressly permitted.
‣ Observe federal/national regulations pertaining to the repair of an electrical device.
‣ Use only original spare parts and accessories from the manufacturer.
Hazardous area
To eliminate danger to persons or the facility when the device is used in the hazardous
area (e.g. explosion protection):‣ Check the nameplate to verify if the device ordered can be put to its intended use in thehazardous area.
‣ Observe the specifications in the separate supplementary documentation that is an
integral part of these instructions.

2.5

Product safety

This measuring device is designed in accordance with good engineering practice to meetstate-of-the-art safety requirements, has been tested, and left the factory in a condition inwhich it is safe to operate.
It meets general safety standards and legal requirements. It also complies with the EC
directives listed in the device-specific EC Declaration of Conformity. The manufacturer
confirms this by affixing the CE mark to the device.

2.6

Safety information for desktop version (option)• The mains plug should only be inserted into a socket with a ground contact.
• The protective effect may not be suspended by an extension cable without a protective
ground.
• Relay outputs: U (max) = 30 Veff (AC) / 60 V (DC)2.7

IT security

The manufacturer only provides a warranty if the device is installed and used as describedin the Operating Instructions. The device is equipped with security mechanisms to protect
it against any inadvertent changes to the device settings.
IT security measures in line with operators' security standards and designed to provide
additional protection for the device and device data transfer must be implemented by theoperators themselves.

Endress+Hauser

9Product description

Ecograph T, RSG35

3

Product description

3.1

Product design

This device is best suited for the electronic acquisition, display, recording, analysis, remotetransmission and archiving of analog and digital input signals.
The device is intended for installation in a panel or cabinet. Operation in a desktop or fieldhousing is possible as an option.

4

Incoming acceptance and productidentification

4.1

Incoming acceptanceOn receipt of the goods, check the following points:• Is the packaging or the content damaged?
• Is the delivery complete? Compare the scope of delivery against the information on yourorder form.

4.2

Scope of delivery

The scope of delivery of the device comprises:• Device (with terminals, as per order)• Panel-mounted device: 2 screw fastening clips• USB cable• Optional: industrial grade SD card (card is located in the SD slot behind the flap on thefront of the device)• "Field Data Manager (FDM)" analysis software on DVD (Essential, Demo or Professionalversion, depending on order)• Delivery note
• Multilanguage Brief Operating Instructions, hard copy

4.3

Product identification

4.3.1

NameplateCompare the nameplate with the following diagram:

10

Endress+Hauser

Ecograph T, RSG35

Installation

1
2
3
4
5
6

35 VA

7

A0019299

1
1
2
3
4
5
6
7

Device nameplate (example)
Device designation
Order code, serial number, extended order code
Power supply, mains frequency
Maximum power consumption
Temperature range
Software version; MAC address
Device approvals

4.3.2

Name and address of manufacturer

Name of manufacturer:

Endress+Hauser Wetzer GmbH + Co. KG

Address of manufacturer:

Obere Wank 1, D-87484 Nesselwang

Model/type reference:

RSG35

4.4

Storage and transport

Compliance with the permitted environmental and storage conditions is mandatory.
Precise specifications are provided in the "Technical data" section. →  72
Please note the following:• Pack the device so that is protected against impact for storage and transport. The
original packaging provides optimum protection.
• The permitted storage temperature is –20 to +60 °C (–4 to +140 °F)

5Installation

5.1

Installation conditions

NOTICE
Overheating due to buildup of heat in the device‣ To avoid heat buildup, always ensure that the device is sufficiently cooled.
The device is designed for use in a panel in non-hazardous areas.
• Ambient temperature range: –10 to +50 °C (14 to 122 °F)• Climate class as per IEC 60654-1: Class B2• Degree of protection: IP65, NEMA 4 at front / IP20 rear of housing

Endress+Hauser

11

Installation

Ecograph T, RSG35

5.1.1

Installation dimensions• Installation depth: approx. 158 mm (6.22 in) for device incl. terminals and fastening
clips• Panel cutout: 138 to 139 mm (5.43 to 5.47 in) x 138 to 139 mm (5.43 to 5.47 in)• Panel thickness: 2 to 40 mm (0.08 to 1.58 in)• Angle of vision: from the midpoint axis of the display, 75° to the left and right, 65° aboveand below• A minimum distance of 15 mm (0.59 in) between the devices must be observed ifaligning the devices in the Y direction (vertically above one another). A minimum
distance of 10 mm (0.39 in) between the devices must be observed if aligning thedevices in the X direction (horizontally beside one another).
• Securing to DIN 43 834

5.2

Mounting the measuring device

Mounting tool: For installation in the panel, all you need is a screwdriver.

17 (0.67)

144 (5.67)

144 (5.67)

141 (5.55)

Y34(1.34)

X

A0019301

2

12Panel mounting and dimensions in mm (inch)

1.

Push the device through the panel cutout from the front. To avoid heat buildup,
maintain a distance of > 15 mm (>0.59 in) from walls and other devices.

2.

Hold the device level and hook the fastening clips into the openings (1 x left, 1 x
right).

Endress+Hauser

Ecograph T, RSG35

Electrical connection

3.

Evenly tighten the screws on the fasting clip using a screwdriver to guarantee asecure seal to the panel (torque 100 Ncm).

5.3

Post-installation check• Is the sealing ring undamaged?
• Does the seal run all around the housing collar?
• Are the threaded rods properly tightened?
• Is the device fixed firmly in the center of the panel cutout?

6

Electrical connection

6.1

Connecting requirements

LWARNING
Danger! Electric voltage!
‣ The entire connection of the device must take place while the device is de-energized.
‣ The mixed connection of safety extra-low voltage and dangerous contact voltage at therelays is not permitted.
‣ Apart from the relays and the supply voltage, only energy-limited circuits according to
IEC/EN 61010-1 may be connected.
Danger if protective ground is disconnected‣ The ground connection must be made before all other connections.
NOTICE
Cable heat load
‣ Use suitable cables for temperatures of 5 °C (9 °F) above ambient temperature.
Incorrect supply voltage can damage the device or cause malfunctions‣ Before commissioning the device, make sure that the supply voltage matches thevoltage specifications on the nameplate.
Check emergency shutdown for device‣ Provide suitable switch or circuit breaker in building installation. This switch must be
provided close to the device (within easy reach) and marked as a circuit breaker.
Protect the device from overload‣ Provide overload protection (nominal current = 10 A) for power cable.
Incorrect wiring may result in the device being destroyed‣ Note terminal designation on the rear of the device.
Energy-rich transients in the case of long signal lines‣ Install suitable overvoltage protection (e.g. E+H HAW562) upstream.

6.2

Connecting instructions

6.2.1

Cable specification

Cable specification, spring terminals
All connections on the rear of the device are designed as pluggable screw or spring
terminal blocks with reverse polarity protection. This makes the connection very quick andeasy. The spring terminals are unlocked with a slotted screwdriver (size 0).

Endress+Hauser

13Electrical connection

Ecograph T, RSG35

Please note the following when connecting:• Wire cross-section, auxiliary voltage output, digital I/O and analog I/O: max. 1.5 mm2(14 AWG) (spring terminals)• Wire cross-section, mains: max. 2.5 mm2 (13 AWG) (screw terminals)• Wire cross-section, relays: max. 2.5 mm2 (13 AWG) (spring terminals)• Stripping length: 10 mm (0.39 in)
No ferrules must be used when connecting flexible wires to spring terminals.
Shielding and grounding
Optimum electromagnetic compatibility (EMC) can only be guaranteed if the systemcomponents and, in particular, the lines - both sensor lines and communication lines - areshielded and the shield forms as complete a cover as possible. A shielded line must be used
for sensor lines that are longer than 30 m. A shield coverage of 90% is ideal. In addition,
make sure not to cross sensor lines and communication lines when routing them. Connect
the shield as often as possible to the reference ground to ensure optimum EMC protection
for the different communication protocols and the connected sensors.
To comply with requirements, three different types of shielding are possible:• Shielding at both ends
• Shielding at one end on the supply side with capacitance termination at the device• Shielding at one end on the supply side
Experience shows that the best results with regard to EMC are achieved in most cases ininstallations with one-sided shielding on the supply side (without capacitance termination
at the device). Appropriate internal device wiring measures must be taken to allow
unrestricted operation when EMC interference is present. These measures have beentaken into account for this device. Operation in the event of disturbance variables as per
NAMUR NE21 is thus guaranteed.
Where applicable, national installation regulations and guidelines must be observed
during the installation! Where there are large differences in potential between theindividual grounding points, only one point of the shielding is connected directly with thereference ground.
If the shielding of the cable is grounded at more than one point in systems without
potential matching, mains frequency equalizing currents can occur. These can damage
the signal cable or significantly impact signal transmission. In such cases the shieldingof the signal cable is to be grounded on one side only, i.e. it may not be connected to
the ground terminal of the housing. The shield that is not connected should be
insulated!

14

Endress+Hauser

Ecograph T, RSG35

Electrical connection

Connecting the measuring device

6.3.1

Terminal assignment on the rear of the device

15Frequency

6.3

9

6

A0019304

3

Terminals on the rear of the device

6.3.2

Supply voltage

Power unit type

Terminal

A0019103

100-230 VAC

24 V AC/DC

6.3.3

L+

N-

PE

Phase L

Zero conductor N

GroundL+

N-

PE

Phase L or +Zero conductor N or -Ground

Relay

Type

Terminal (max. 250 V, 3 A)

A0019103

Alarm relay 1

Endress+Hauser

R11

R12

R13

Changeover
contact

Normally
closed contact(NC) 1)

Normally open
contact (NO) 2)

15

Electrical connection

Ecograph T, RSG35

Type

Terminal (max. 250 V, 3 A)

A0019103

Relay 2 to 6

1)
2)

Rx1

Rx2

Switching contact

Normally open
contact (NO 2))

NC = normally closed (breaker)
NO = normally open (maker)

6.3.4

Digital inputs; auxiliary voltage output

TypeTerminal

Digital input
1 to 6

D11 to D61

GND1

Digital input 1 to 6(+)

Ground (-) for digital
inputs 1 to 6

A0019103

Auxiliary
voltage
output, notstabilized,
max. 250mA

6.3.5

24V Out -24V Out +- Ground+ 24V (±15%)Analog inputs

The first digit (x) of the two-digit terminal number corresponds to the associated channel:
Terminal

x1
x2
x3
x4
x5
x6

Chx

Type

A0019303

x1

x2

x3

x4

Current/pulse/frequency
input 1)
Voltage > 1V(+)

Voltage ≤ 1V

16

(A)

Resistance thermometer
RTD (3-wire)

(A)

x6

(+)

(-)
(-)

(+)

Resistance thermometer
RTD (2-wire)

x5

(-)
(B)

b (sense)(B)

Endress+Hauser

Ecograph T, RSG35

Electrical connection

Terminal

x1
x2
x3
x4
x5
x6

Chx

Type

A0019303

Resistance thermometer
RTD (4-wire)(A)a (sense)

Thermocouples TC

(B)

(+)

(-)

If a universal input is used as a frequency or pulse input and the voltage is >2.5 V, a series resistor must be
used in series connection with the voltage source. Example: 1.2 kΩ series resistor at 24 V

Connection example: Auxiliary voltage output as transmitter
power supply for 2-wire sensors

+

+

6.3.6

-

1)

b (sense)

I
3

Y

_

2
1
+

+

I
Y

_

Frequency

Out: max. 250 mA

A00202594
1
2
3

Endress+Hauser

Connection of the auxiliary voltage output when using as a transmitter power supply for 2-wire sensors in
the current measuring range. (When connecting channel CH3-12, see pin assignment CH1-2.)
Sensor 1 (e.g. Cerabar from Endress+Hauser)
Sensor 2
External indicator (optional) (e.g. RIA16 from Endress+Hauser)

17

Electrical connection

Ecograph T, RSG35

Connection example: Auxiliary voltage output as transmitter
power supply for 4-wire sensors
+

Y

-

I

3

24V

-

2

Y

+

24V

-

I

+

_

+

+

6.3.7

_

+
-

1

Frequency

Out: max. 250 mA

A002026051
2
3Connection of the auxiliary voltage output when using as a transmitter power supply for 4-wire sensors in
the current measuring range. (When connecting channel CH3-12, see pin assignment CH1-2.)
Sensor 1 (e.g. temperature switch TTR31 from Endress+Hauser)
Sensor 2
External indicator (optional) (e.g. RIA16 from Endress+Hauser)

6.3.8

Option: RS232/RS485 interface (rear of device)

Use shielded signal lines for serial interfaces!
A combined RS232/RS485 connection is available on a shielded SUB D9 socket at the rearof the device. This can be used for data transfer and to connect a modem. For
communication via modem, we recommend an industrial modem with a watchdog
function.

18

Endress+Hauser

Ecograph T, RSG35

Electrical connection

Termination
resistor(typical 120 Ω)

Further units

To modem: Cable with
25 pol. Sub-D socket

To modem: Cable with
9 pol. Sub-D socket

To PC: Cable with
25 pol. Sub-D
plug

To PC: Cable with
9 pol. Sub-D plug
RxD - 3
GND - 5TxD - 2

A0019305-EN

Type

Pin of the SUB-D9 socket
1 2

RS232
assignment

TxD (data
output)3
RxD (data
input)

RS485
assignment

4 5

6 7 8

9

GND
GND

RxD/TxD –RxD/TxD +Unoccupied connections should be left empty.
Maximum cable length:
RS232: 2 m (6.6 ft)
RS485: 1000 m (3280 ft)

Only one interface can be used at any one time (RS232 or RS485).

6.3.9

Ethernet connection (rear of device)

The Ethernet interface can be used to integrate the device via a hub or switch into a PC
network (TCP/ IP Ethernet). A standard patch cable (e.g. CAT5E) can be used for theconnection. Using DHCP, the device can be fully integrated into an existing network
without the need for additional configuration. The device can be accessed from every PC in
the network.
• Standard: 10/100 Base T/TX (IEEE 802.3)• Socket: RJ-45• Max. cable length: 100 m• Galvanic isolation; test voltage: 500 V

Endress+Hauser

19Electrical connection

Ecograph T, RSG35

Meaning of the LEDs
Beneath the Ethernet connection (see rear of device) there are two light emitting diodes
which indicate the status of the Ethernet interface.
• Yellow LED: link signal; is lit when the device is connected to a network. If this LED is not
lit, communication is not possible.
• Green LED: Tx/Rx; flashes irregularly if the device is transmitting or receiving data.

6.3.10

Option: Ethernet Modbus TCP slave

The Modbus TCP interface is used to connect to higher-ranking SCADA systems (Modbus
master) to transmit all measured values and process values. Up to 12 analog inputs and 6
digital inputs can be transmitted via Modbus and stored in the device. The Modbus TCP
interface is physically identical to the Ethernet interface.

6.3.11

Option: Modbus RTU slave

The Modbus RTU (RS485) interface is galvanically isolated (test voltage: 500 V) and isused to connect to higher-ranking systems to transmit all measured values and process
values. Up to 12 analog inputs and 6 digital inputs can be transmitted via Modbus andstored in the device. Connection is via the combined RS232/RS485 interface.
Modbus TCP and Modbus RTU cannot be used at the same time.

6.3.12Connections at front of device
11

10

9

12
8

7

6 5 4

3

2

1
A0019501

6

Front of device with open flap

1
Navigator
2
LED at SD slot. Orange LED lights up or flashes when the device writes to or reads from the SD card.
3
Slot for SD card
4
USB B socket "Function" e.g. to connect to PC or laptop
5Green LED lit: power supply present
6
USB A socket "Host" e.g. for USB memory stick or external keyboard
7-12 For a description of the display formats, see the "Operability" section

20

Endress+Hauser

Ecograph T, RSG35

Electrical connection

USB connection type A (host)A USB 2.0 port is available on a shielded USB A socket at the front of the device. A USB
stick, for example, can be connected to this interface as a storage medium. An external
keyboard or USB hub may also be connected.
USB connection type B (function)
A USB 2.0 port is available on a shielded USB B socket at the front of the device. This canbe used to connect the device for communication with a laptop, for example. →  32
USB 2.0 is compatible with USB 1.1 or USB 3.0, i.e. communication is possible.
Information on USB devices
The USB devices are detected by the "plug-and-play" function. If several devices of the same
type are connected, only the USB device that was connected first is available. Settings for
the USB devices are made in the setup. A maximum of 8 external USB devices (incl. USBhub) can be connected if they do not exceed the maximum load of 500 mA. If overloaded,
the corresponding USB devices are automatically disabled.
Requirements with regard to an external USB hub
The USB devices are detected by the "plug-and-play" function. If several devices of the same
type are connected, only the USB device that was connected first is available. Settings for
the USB devices are made in the setup. A maximum of 8 external USB devices (incl. USBhub) can be connected if they do not exceed the maximum load of 500 mA. If overloaded,
the corresponding USB devices are automatically disabled.
Requirements with regard to the USB stick
There is no guarantee that all manufacturers' USB sticks will function faultlessly. That is
why an industrial grade SD card is recommended to ensure the reliable recording of data.
→  70The USB stick must be formatted to FAT or FAT32. NTFS format is not readable. The
system supports only USB sticks with max. 32 GB.
The USB stick must not be connected to the device via a USB hub. Interference fromother USB devices may result in data loss.
Requirements with regard to an external USB keyboard
The system only supports keyboards which can be addressed using generic drivers (HID
keyboard - Human Interface Device). Special keys are not supported (e.g. Windows keys).
Users can only enter characters that are available in the input character set of the device.
All unsupported characters are rejected. It is not possible to connect a wireless keyboard.
The following keyboard layouts are supported: DE, CH, FR, USA, USA International, UK, IT.
See setting under "Setup -> Advanced setup -> System -> Keyboard layout".
Requirements for the SD card
Industrial grade SD-HC cards with max. 32 GB are supported.
Use only the industrial grade SD cards described in the "Accessories" section of the
Operating Instructions. These have been tested by the manufacturer and are
guaranteed to function faultlessly in the device. →  70
The SD card must be formatted to FAT or FAT32. NTFS format is not readable.

Endress+Hauser

21

Electrical connection

Ecograph T, RSG35

6.4

22

Post-connection check

Device condition and specifications

Notes

Are cables or the device damaged?

Visual inspection

Electrical connection

Notes

Does the supply voltage match the specifications on the nameplate?

-

Are all terminals firmly engaged in their correct slot?

-

Are the mounted cables strain-relieved?

-

Are the power supply and signal cables correctly connected?

See connection diagram andrear of device.

Endress+Hauser

Ecograph T, RSG35

Operation options

7

Operation options

7.1

Overview of operation options

The device can be operated directly onsite with the navigator and USB keyboard/mouse or
via interfaces (serial, USB, Ethernet) and operating tools (Web server); FieldCare/
DeviceCare configuration software).

7.2

Structure and function of the operating menu
The layout and structure of the operating menu can differ slightly in parts on the Web
server.

Endress+Hauser

23

Operation options

Ecograph T, RSG35

7.2.1

Operating menu for operators and maintenance personnel

Operator

Operation

Setup

Change group

SD card/USB stick

Search in trace

Lock operation

History

Change display mode

Signal analysis

Adjust brightness

Change date/time

Advanced setup

SystemDate/time setup
Security
External memory
Messages
Screen saver
Device options,...

Inputs

Universal inputs

MaintenanceDigital inputs

OutputRelaiy 1 - 6

Communication

Ethernet
Serial interface

Modbus Slave

Application

Maths
Signal analysis
LimitsSignal groups
E-mailDiagnostics

Diagnostics list

WebDAV Client

Event logbook
Device information
Measured Values
Outputs
Simulation
Initialize modem
A0019594-EN

24

Endress+Hauser

Ecograph T, RSG35

Operation options

7.2.2

Operating menu for experts

ExpertEnter Access code
Direct Access
System(see before)+Clear memory+Expert functions

Inputs(see before)+Expert functions

ExpertOutputs(see before)+Expert functions

Communication(see before)+Expert functions

Application(see before)+Expert functions

Diagnostics

Diagnosis list
Event logbook
Device information

Simulation

A0019596-EN

7.2.3

Submenus and users

Certain parts of the menu are assigned to certain user roles. Each user role corresponds totypical tasks within the lifecycle of the device.
User role

Typical tasks

Menu

Content/meaning

Operator

Tasks during operation:• Configuration of the display.
• Reading measured values.

"Operation"

Contains all the parameters that are required inongoing operation: configuration of the measured value
display (displayed values, display format, etc.).

MaintenanceCommissioning:• Configuration of the measurement.
• Configuration of data processing.

"Setup"Contains all parameters for commissioning:• Change date/time• "Advanced setup" submenu
Contains additional submenus and parameters:• System: Basic settings required for operating thedevice.
• Inputs: Settings for analog and digital inputs.
• Outputs: Setup required only if outputs (e.g.
relays) are to be used.
• Communication: Settings required if the USB,
RS232, RS485 or Ethernet interface of the device
is used (PC operation, serial data read-out, modem
operation, etc.).
• Application: Various application-specific settings(e.g. group settings, limit values, etc.).
Once values have been set for these parameters, themeasurement should generally be completely
configured.

Endress+Hauser

25

Operation options

User role

Expert26

Ecograph T, RSG35

Typical tasks

Menu

Content/meaning

Troubleshooting:• Diagnosing and eliminating process errors.
• Interpretation of device error messages andcorrecting associated errors.

"Diagnostics"

Contains all parameters for detecting and analyzing
errors:• Diagnosis list
All diagnosis messages are listed in chronologicalorder.
• Event logbook
Events such as alarm set point infringement andpower failure are listed in the correct time sequence.
• Device information
Displays important device information (e.g. serial
number, firmware version, device options for
hardware and software, memory information, etc.).
• Measured values
Display of current measured values of the device.
• Outputs
Displays the current status of the outputs e.g. switch
status of relay outputs.
• Simulation
Various functions/signals can be simulated for test
purposes here.
Note: In Simulation mode, normal recording of themeasured values is interrupted and the intervention
is logged in the event log.
• Initialize modem
Initializes the modem connected to the serialinterface (for automatic call answering).

Tasks that require detailed knowledge of thefunction of the device:• Commissioning measurements under difficult
conditions.
• Optimal adaptation of the measurement todifficult conditions.
• Detailed configuration of the communicationinterface.
• Error diagnostics in difficult cases.

"Expert"Contains all the parameters of the device (including
those that are already contained in one of the othersubmenus). The expert menu is protected by a code.
Factory setting: 0000. This menu is structured
according to the function blocks of the device:• "System" submenu
Contains all higher-level device parameters that do
not pertain either to the measurement or measured
value communication.
• "Inputs" submenu
Contains all parameters for configuring the analog
and digital inputs.
• "Output" submenu
Contains all parameters for configuring the outputs(e.g. relays).
• "Communication" submenu
Contains all parameters for configuring thecommunication interfaces.
• "Application" submenu
Contains all parameters for configuring applicationspecific settings (e.g. group settings, limit valuesetc.).
• "Diagnostics" submenu
Contains all parameters to detect and analyze
operational errors.

Endress+Hauser

Ecograph T, RSG35

Operation options

7.3

Measured value display and operating elements
11

10

9

12
8

7

6 5 4

3

2

1
A0047011

7

Front of device with open flap

Item
No.

Operating function (display mode = display of measured values)(Setup mode = operation in the Setup menu)

1"Navigator": Jog/shuttle dial for operating with additional press/hold function.
In Display mode: Turn the dial to switch between the various signal groups. Press the dial to display
the main menu.
In Setup mode or in a selection menu: Turn the dial anticlockwise to move the bar or the cursor up or
to the left, changes the parameter. Turning clockwise moves the bar or cursor down or to the right,changes the parameter.
Press briefly (<2 sec.) = Select highlighted function, parameter change starts (ENTER key).
Access online help: Press and hold navigator (>3 sec.) to show information on the selected function.
To quit the menu immediately, press and hold "Back" (>3 sec.) in the navigator. The devices
switches to Display mode.

2

LED at SD slot. Orange LED lit when the device writes to the SD card or reads it.
Do not remove the SD card if the LED is lit! Risk of data loss!

3

Slot for SD card

4

USB B socket "Function" e.g. to connect to PC or laptop

5Green LED lit: power supply present

6

USB A socket "Host" e.g. for USB memory stick or external keyboard

7

In Display mode: Alternating status display (e.g. set zoom range) of the analog or digital inputs in thecorresponding color of the channel.
In Setup mode: Different information can be displayed here depending on the display type.

8

In Display mode: Window for measured value display (e.g. curve display).
In Setup mode: Display of operating menu

Endress+Hauser

27

Operation options

Ecograph T, RSG35

Item
No.

Operating function (display mode = display of measured values)(Setup mode = operation in the Setup menu)

9

In Display mode: Current group name, analysis type
In Setup mode: Name of the current operating item (dialog title)

10

In Display mode: Current date/time is displayed
In Setup mode: --11

In Display mode: Alternating display indicating the percentage space on the SD card or USB stick thathas already been used.
Status symbols are also displayed in alternation with the memory information (see the following
table).
In Setup mode: The current "direct access" operating code is displayed

12

In Display mode: The current measured values are displayed, and in the event of an error/alarm
condition the status is displayed. In the case of counters, the type of counter is displayed as a symbol(see the following table).
point has limit value status, the corresponding channel identifier is highlighted
 inIf aredmeasuring(quick detection of limit value violations). During a limit value violation and device
operation, the acquisition of measured values continues uninterrupted.

7.4

Display representation of symbols used in operation

Item
No.

Function

8,12

Symbols for counters:

8, 12

Description

Σ1Intermediate analysis / external analysis

ΣD

Daily analysis

ΣM

Monthly analysis

ΣY

Annual analysis

ΣTotalizer

Channel-related symbols:
Violation of lower limit value
Violation of upper limit value or limit value on counter
Violation of upper and lower limit values at the same time
"Out of specification"e.g. input signal too high/low
Error message "Failure detected"
An operating error has occurred. The measured value is no longer valid (e.g.
a channel not displayed in the current group is defective).
"Maintenance required"
Maintenance is required. The measured value remains valid.
Error, measured value not displayed.
Possible causes: Sensor / input error, line break, invalid value, input signal
too high/low

11

Symbol for status signals:"Device locked"
Setup is locked via a control input. Disable setup lock via a control input.
"Out of specification"
The device is being operated outside its technical specifications (e.g. during
startup or cleaning).
"Function check"
The device is in Service mode.

28

Endress+Hauser

Ecograph T, RSG35

Operation options

Item
No.

Function

Description"Maintenance required"
Maintenance is required. The measured value remains valid.
Error message "Failure detected"
An operating error has occurred. The measured value is no longer valid (e.g.
a channel not displayed in the current group is defective).
"External communication"
The device is communicating externally (e.g. via Modbus).

SIM"Simulation"
Simulation is active.

7

7.4.1

"Historical data"
Historical data are currently shown on screen.

Symbols in operating menus
Symbol for setup
Symbol for diagnostics
Symbol for expert setup
Back
The function "Back" appears at the end of each menu/submenu.
Press "Back" briefly to go up one level in the menu structure.
Press and hold "Back" (>3 sec.) to quit the menu immediately. The devices switches to Display
mode.

7.4.2

Symbols in the event logbook
Setup changes

Power onPower off
Limit value on
Limit value off

1Digital on (on/off message)0

Digital off (on/off message)
Service
Texts saved / comments added
Acknowledging message
Back
Continue searching

7.5

Entering text and numbers (virtual keyboard)

A virtual keyboard is available for entering text and numbers. This is opened automatically
if needed. The appropriate character is selected by turning and pressing the navigator.
The following characters are available for entering customized text:

Endress+Hauser

29

Operation options

Ecograph T, RSG35

0-9 a-z A-Z = + - * / \ 2 3 ¼ ½ ¾ ( ) [ ] < > { } I ? ! ` " ' ^ % ° . , : _ µ & # $ € @ § £ ¥ ~←Jump one position to the left.
If this symbol is selected, the cursor jumps one position to the left.

→

Jump one position to the right.
If this symbol is selected, the cursor jumps one position to the right.

←xDelete backwards.
If this symbol is selected, the character to the left of the cursor position is deleted.

x→Delete forwards.
If this symbol is selected, the character to the right of the cursor position is deleted.
Delete all.
If this symbol is selected, the entire entry is deleted.
Reject entry.
If this symbol is selected, the entry is rejected and you quit editing mode. The previously set text
remains.
Accept entry.
If this symbol is selected, the entry is applied at the position specified by the user, and you quit
editing mode.

7.6

Channel color assignment

Channel color assignment is performed in the main menu under "Setup -> Advanced
setup -> Application -> Signal groups -> Group x". 8 predefined colors are available pergroup and can be assigned to the desired channels.

7.7

Access to the operating menu via the local display

Using the "Navigator" (jog/shuttle dial with additional press/hold function), all settings canbe made directly onsite at the device.

7.8

Device access via operating tools

7.8.1

Field Data Manager (FDM) analysis software (SQL databasesupport)The PC analysis software offers external, centralized data management with visualization
for recorded data. The analysis software enables the complete archiving of all measuring
point data e.g. measured values, diagnostic events and protocols. The analysis software
stores data in an SQL database. The database can be operated locally or in a network(client/server). Access is via RS232/RS485, USB or Ethernet interface (network).
Function scope:• Export of saved data (measured values, analyses, event log)
• Visualization and processing of saved data (measured values, analyses, event log)
• Safe archiving of exported data in a SQL database
The following versions of the software are available:• Essential version (free, with limited functionalities)• Professional version (see Accessories →  70)• Demo version (time-limited Professional version)
An "Essential" version of the analysis software is supplied with the device.
For details, see the Operating Instructions on the analysis software DVD provided.

30

Endress+Hauser

Ecograph T, RSG35

Operation options

7.8.2

Web server

A Web server is integrated into the device. This makes the current measured values of thedevice available in real time. Access is via an Ethernet interface from a PC in the network
via the standard browser. The installation of additional software is not required.
Alternatively, the Web server can be operated in a point-to-point connection (Ethernet
over USB) with a standard USB cable via the USB B port. →  32
The Web server offers the following range of functions:• Display of current and historical data and measured value curves via a standard Webbrowser →  42
• Easy configuration without additional installed software →  35
• Remote access to device and diagnostic information

7.8.3

OPC server (optional)

The OPC server makes it possible to access data on the device. These data are madeavailable to OPC clients in real time. The OPC server meets the requirements of the OPC
specifications regarding the supply of data to an OPC client. Access is via RS232/RS485,
USB or Ethernet interface (network). Communication takes place using automatic device
detection; the operator does not need to make any additional settings. The OPC server
enables the flexible and powerful exchange of data and is easy and convenient to use.
The following instantaneous values can be provided:• Analog channels• Digital channels• Mathematics• Totalizer
For details, see Operating Instructions BA00223R/09/xx

7.8.4

FieldCare/DeviceCare configuration software

Function scope
The configuration software is an FDT/DTM-based system asset management tool. It canconfigure all smart field devices in a system and helps you manage them. By using thestatus information, it is also a simple but effective way of checking their status andcondition. Access is via USB or Ethernet interface (network).
Typical functions:• Device configuration• Loading and saving device data (upload/download)• Documentation of the measuring point
Download at: www.endress.com/download
Overview of device description files (DTM)
Information and files are available free of charge at:
See online at: www.de.endress.com/fieldcare

Endress+Hauser

31

System integration

Ecograph T, RSG35

8

System integration

8.1

Integrating the measuring device in the system

8.1.1

General notes

The device has (optional) fieldbus interfaces for exporting process values. Measured values
and statuses can also be transmitted to the device via fieldbus.
Note: Counters cannot be transferred.
Depending on the bus system, alarms or faults occurring during data transmission aredisplayed (e.g. status byte).
The process values are transferred in the same units that are used to display the values on
the device.

8.1.2

Ethernet

Setup → Advanced setup → Communication → Ethernet
The IP address can be entered manually (fixed IP address) or assigned automatically using
DHCP.
The port for data communication is preset to 8000. The port can be changed in the Expert→ Communication → Ethernet menu.
The following functions are implemented:• Data communication with PC software (analysis software, configuration software, OPC
server)• Web server
The following connections are possible at the same time:• 1x Port 8000 (configuration software, OPC server or analysis software)• 1x Port 8002 (OPC server only)• 4x Modbus slave TCP• 5x Web server
Ports can be changed!
As soon as the maximum number of connections has been reached, new connection
attempts are blocked until an existing connection has been terminated.

8.1.3

Web server with "Ethernet over USB" function

To provide easy and efficient access via Web server for operation, setup andcommissioning, the USB-B interface can be switched to the "Ethernet over USB" mode.
Ethernet communication is established here via the USB interface. This has the advantagethat the Ethernet interface in the case of commissioning laptops, for example, does notneed to be reconfigured (IP address, ports, etc.). Instead, a standard USB cable can be used

32

Endress+Hauser

Ecograph T, RSG35

System integration

to establish a point-to-point connection. The Web server itself retains its complete rangeof functionality.
Important information:• Do not connect several devices via USB to a laptop/PC at the same time
• The PC should not be connected to "Ethernet via USB" (USB-B connector on front
panel of device) and a standard Ethernet network (RJ45 connector on rear panel ofdevice) simultaneously if both connections use the same address parameters.
• Driver software (EH ECM device) must be installed on the PC.
• The "Ethernet over USB" mode is not a gateway, i.e. the Ethernet/RJ45 and USB
networks are separate from each other (no two-way access).
• The USB cable should be disconnected from the device for at least 10 s before being
reconnected (prevention of errors due to response time of system)• It takes at least 10 s before the USB driver is activated by Windows andcommunication with the device is possible.
• Initialization of the USB interface takes place when the PC and device are connected
via a USB cable.
Supported services/functions
The following services/functions are provided via the USB interface:• Web server• CDI TCP (port 8000)• WebDAV server
All other services/functions are available only by Ethernet via RJ45!
Driver installation at PC end
To use the Web server via USB, a driver must be installed once at the PC end.
1.

The driver can be found under "..\Drivers\USB_ECM" on the Field Data Manager(FDM) software DVD supplied.

2.

Alternatively: Download "USB_ECM.zip" at www.endress.com/downloads

3.

Execute the "setup.exe" and follow the instructions.

The necessary drivers are installed.
From version V1.04.01 of the Field Data Manager (FDM) software provided, thedriver is automatically installed with the PC software.
The driver can be installed directly from the DVD without first having to copy the files
to the PC.
Changing the USB-B mode to "Ethernet over USB" (panel version)Communication can be established only if the device has been configured accordingly.
1.

Supply power to the device, restart and wait for start screen.

2.

Open dropdown menu under "Setup → Advanced setup → Communication →Function USB-B".

3.

"Always USB" option: USB-B is always set to Standard USB.

4.

"Always Ethernet over USB" option: Ethernet over USB is always set.

5.

"By user input" option: A prompt for the mode (option) appears once a USB cable isplugged in.

The device is now ready to establish a connection.
Establishing communication

Endress+Hauser

1.

Using a standard USB cable, connect the device at the USB-B interface to any of the
PC's USB interfaces.

2.

With "Always Ethernet over USB" function: The Web server can be started
immediately.
33

Commissioning

Ecograph T, RSG35

3.

With "By user input" function: Once the USB cable has been plugged into the device,the "Select USB function" dropdown menu appears. Then select "Ethernet over USB" on
the device. The device returns the IP address.

4.

Open the browser, enter http://192.168.1.212 and follow the instructions fromsection →  43

The device is now ready to communicate with the Web server.
The IP address of the Web server via USB is set permanently to http://192.168.1.212
If the USB cable is already connected to a PC when the device starts up, there is noprompt even if the "By user input" option is selected. Instead, the previously selectedfunctionality is used.

8.1.4

Modbus RTU/TCP slave

The device can be connected to a Modbus system via RS485 or Ethernet interface. The
general settings for the Ethernet connection are made in the Setup → Advanced setup →Communication → Ethernet menu. Modbus communication is configured in the Setup →Advanced setup → Communication → Modbus Slave menu.
Up to 12 analog inputs and 6 digital inputs can be transmitted via Modbus and stored in
the device.
Menu position

RTU (RS485)

Ethernet

Device address:

1 to 247

IP address manual or automatic

Baud rate:

9600/19200/38400/57600/115200-

Parity:Even/Odd/None-

Stop bits:

1/2

-

Port:

-

502

Transfer of values
The actual Modbus TCP protocol is located between layer 5 to 6 in the ISO/OSI model.
To transfer a value, 3 registers of 2 bytes each (2-byte status + 4-byte float) or 5 registersof 2 bytes each (2-byte status + 8-byte double) are used.
For more information about Modbus, see the supplementary documentation.

9

Commissioning

9.1

Function check

Perform the following checks prior to commissioning:• "Post-mounting check" checklist →  13.
• "Post-connection check" checklist →  22.

9.2

Switching on the measuring device

After the operating voltage is applied, the green LED lights up and the device is ready for
operation.
If you are commissioning the device for the first time, program the setup as described in
the following sections of the Operating Instructions.
34

Endress+Hauser

Ecograph T, RSG35

Commissioning

If you are commissioning a device that is already configured or preset, the device starts
measuring immediately as defined in the settings. The values of the channels currently
activated are shown on the display.
Remove the protective film from the display as this would otherwise affect thereadability of the display.

9.3

Setting the operating language

Factory setting: English or ordered local language
Calling the main menu, configuring the operating language:
1.

Press the navigator

2.

The main menu appears on the display with the "Sprache/Language" option

3.

To change the set language: Press the navigator, turn the navigator to select thedesired language and press the navigator to apply the change.

4.

Use "Back" or "ESC" to quit the main menu

The operating language has been changed.
The function"Back" appears at the end of each menu/submenu.

Press "Back" briefly to go up one level in the menu structure.
To quit the menu immediately and return to the measured value display, press andhold "Back" (>3 sec.). The changes made are accepted and saved.

9.4

Configuring the measuring device (Setup menu)

Access to the setup is enabled when the device leaves the factory and can be locked invarious ways e.g. by entering a 4-digit access code or via role-based password protection.
When locked, basic settings can be checked but not changed. The device can also be put
into operation and configured via the PC.
Device configuration options:• Setup directly at the device (panel-mounted device only)• Setup via SD card or USB stick by transferring the parameters stored on it• Setup via Web server using Ethernet or Ethernet over USB• Setup via FieldCare/DeviceCare configuration software

9.4.1

Step-by-step: to the first measured value

Procedure and necessary settings:

Endress+Hauser

1.

Check the date/time in the main menu under "Setup" and set it if necessary

2.

Make settings for the interfaces and communication in the main menu under "Setup-> Advanced setup -> Communication"

3.

Create universal or digital inputs in the main menu under "Setup -> Advanced setup-> Inputs -> Universal inputs/Digital inputs": Add input: select "Universal inputx" or "Digital input x" with which the input signal should be detected. Then selectand configure the new input that has been created.

4.

Activate relays or analog outputs (optional) in the main menu under "Setup ->Advanced setup -> Outputs"

5.

Assign activated inputs to a group in the main menu under "Setup -> Advanced
setup -> Application -> Signal groups -> Group x"

6.

Use "Back" or "ESC" to quit the main menu. The changes made are accepted and saved.

35

Commissioning

Ecograph T, RSG35

The device is in the measured value display mode and displays the measured values.

9.4.2

Step-by-step: set or delete the limit values

Procedure for setting limit values:
1.

Open the limit values in the main menu under "Setup -> Advanced setup ->Application -> Limits"2.

Add a limit value: select "Yes"3.

Select and configure "Limit value x"

4.

Use "Back" or "ESC" to quit the main menu. The changes made are accepted and saved.

The device is in the measured value display mode and displays the measured values.
Procedure for deleting limit values:
1.

Open the limit values in the main menu under "Setup -> Advanced setup ->Application -> Limits"

2.

Delete a limit value: select "Yes"3.

Select the limit value to be deleted from the list

4.

Use "Back" or "ESC" to quit the main menu. The changes made are accepted and saved.

The device is in the measured value display mode and displays the measured values.

9.4.3

Device setup

You can open the main menu by pressing the navigator during operation. Turn thenavigator to navigate through the available menus. When the desired menu is displayed,press the navigator to open the menu.
In the "Setup" menu and in the "Advanced setup" submenu, you will find the mostimportant settings for the device:
Parameter

Possible settings

Description

Change date/time

UTC time zone
dd.mm.yyyy hh:mm:ss

Change the date and time.

Advanced setup

Advanced settings for the device e.g. system settings,
inputs, outputs, communication, application etc.

SystemBasic settings that are needed to operate the device,(e.g. date/ time, security, memory management,
messages, etc.)

InputsSettings for the analog and digital inputs.

Outputs

Settings only required if outputs (e.g. relays or
analog outputs) are to be used.

Communicati
on

Settings required if the USB, RS232/RS485 or
Ethernet interface of the device is to be used (PCoperation, serial data export, modem operation, etc.).
different interfaces (USB, RS232/RS485,
 The
Ethernet) can be operated in parallel. However,simultaneous use of the RS232 and RS485
interface is not possible.

Application

Various application-specific settings (e.g. groupsettings, limit values, etc.).

For a detailed overview of all the operating parameter, please refer to the Appendix of
the Operating Instructions. →  88

36

Endress+Hauser

Ecograph T, RSG35

Commissioning

9.4.4

Setup via SD card or USB stick

An existing device configuration ("Setup data" *.DEH) from another Ecograph T RSG35 orfrom FieldCare/DeviceCare can be uploaded directly to the device.
Import new setup directly at the device: The function used to load the setup data can befound in the main menu under "Operation -> SD card (or USB stick) -> Load setup ->Select directory -> Next".

9.4.5

Setup via Web server

To configure the device via the Web server, connect the device to a PC via Ethernet (orEthernet over USB).
Please observe the information and communication settings for Ethernet (or Ethernet overUSB) and the Web server under →  32.
To configure the device via a Web server, Administrator or Service authentication isrequired. ID and password administration is performed in the main menu under"Setup -> Advanced setup -> Communication -> Ethernet -> Configuration Web
server -> Authentication".
ID default value: admin; Password: admin
Note: The password should be changed during commissioning!
Establishing a connection and setup
Procedure for setting up a connection:
1.

Connect the device to the PC via Ethernet (or Ethernet over USB).

2.

Start the browser on the PC; enter the IP address: http://<IP address> to open the
Web server for the device. Note: Leading zeros in IP addresses must not be entered(e.g. enter 192.168.1.11 instead of 192.168.001.011).

3.

Enter the ID and password, and confirm each by clicking "OK" (see also the "Webserver" section of the Operating Instructions →  43)

4.

The Web server shows the instantaneous value display of the device. In the Web
server taskbar, click "Menu -> Setup -> Advanced setup".

5.

Start the configuration

Continue with device configuration in accordance with the Operating Instructions for thedevice. The complete Setup menu i.e. all of the parameters listed in the Operating
Instructions, can also be found on the Web server. After configuration, accept the setup
with "Save settings".
Procedure to establish a direct connection via Ethernet (point-to-point connection):→  42NOTICE
Undefined switching of outputs and relays‣ When configuring using a Web server, the device may adopt undefined states! This
may result in the undefined switching of outputs and relays.
An existing device configuration ("Setup data" *.DEH) from another Ecograph T RSG35
or from FieldCare/DeviceCare can be uploaded directly to the device via the Web
server.
Procedure for uploading a new setup via the Web server:Endress+Hauser

1.

Establish a connection to the device via the Web server →  37

2.

Click "Data management -> Import device settings" in the Web server function bar

3.

Select the setup file and press "OK" to confirm
37Commissioning

Ecograph T, RSG35

4.

The file is transferred, checked and accepted5.

Once the device settings are accepted, information to this effect is displayed in the
Web server.

9.4.6

Setup via FieldCare/DeviceCare configuration software

To configure the device using the configuration software, connect the device to your PC via
USB or Ethernet.
Download at: www.endress.com/download
Establishing a connection and setup
Continue with device configuration in accordance with the Operating Instructions for thedevice.
The complete Setup menu, i.e. all the parameters listed in the Operating Instructions, canalso be found in the configuration software.
NOTICE
Undefined switching of outputs and relays‣ During configuration using the configuration software, the device may assume
undefined statuses! This may result in the undefined switching of outputs and relays.

9.5

Advanced settings (Expert menu)

You can open the main menu by pressing the navigator during operation. Turn thenavigator to navigate to the "Expert" menu. Press the navigator to open the menu.
The Expert menu is protected by the code "0000". If an access code is set up under"Setup -> Advanced setup -> System -> Security -> Protected by -> Access code",
this must be entered here.
The "Expert" menu contains all of the device settings:
Parameter

Possible settings

Description

Direct access

000000-000

Direct access to parameters (fast access)

SystemBasic settings that are needed to operate the device,(e.g. date/ time, security, memory management,
messages, etc.)

InputsSettings for the analog and digital inputs.

Outputs

Settings only required if outputs (e.g. relays or
analog outputs) are to be used.

Communication

Settings required if the USB, RS232/RS485 or
Ethernet interface of the device is to be used (PCoperation, serial data export, modem operation, etc.).
different interfaces (USB, RS232/RS485,
 The
Ethernet) can be operated in parallel. However,simultaneous use of the RS232 and RS485
interface is not possible.

Application

Define various application-specific settings (e.g.
group settings, limit values, etc.).

Diagnosis

Device information and service functions for a quick
device check.

A detailed overview of all the operating parameters is provided in the appendix at theend of the Operating Instructions. →  88

38

Endress+Hauser

Ecograph T, RSG35

Commissioning

9.6

Configuration management

You can save the setup data ("Configuration") to an SD card or a USB stick, to a PC
drive via the Web server, or store them in a database using the configuration
software. This allows additional devices to be configured very easily using the samesettings.
Save setup: The function used to save the setup files can be found in the main menu under"Operation -> SD card (or USB stick) -> Save setup".

LCAUTION
If the SD card or USB stick are removed directly:
Risk of data loss on SD card or USB stick‣ To remove the SD card or the USB stick, always select "Operation -> SD card (or USB
stick) -> Safe removal" in the main menu!
Procedure for saving a setup via the Web server:
1.

Establish a connection to the device via the Web server →  37

2.

Click "Data management -> Save device settings" in the Web server function bar

3.

Select the setup file4.

Transfer the file5.

Verify and accept

6.

Once the device settings are accepted, information to this effect is displayed in the
Web server.
The function for saving the setup data must be enabled at the device for the Web
server under "Setup -> Advanced setup -> Communication -> Ethernet ->Configuration Web server; Setup -> Yes".

9.7

Simulation

Various functions/signals can be simulated for test purposes here.
NOTICE
Selecting simulation: Simulation of the relays and the WebDAV client can be found in
the main menu under "Diagnostics -> Simulation". The simulation of the measured
values can be found in the main menu under "Expert -> Diagnostics -> Simulation".
Only the simulated values are recorded during simulation. The simulation is recorded in
the event logbook.
‣ Do not start simulation if measured value recording must not be interrupted!

9.8

Access protection and security concept

To protect the setup from unauthorized access following commissioning, there are many
options to ensure access protection to the setup settings and the user entries. Access andauthorizations can be configured and assigned passwords.
The user of the device is responsible for access protection and the security concept. In
addition to the device functions listed, user policies and procedures, in particular,
must also be applied (e.g. password allocation, password sharing, physical accessbarriers, etc.).
The following protection options and functionalities are available:• Protection per control input• Protection via access code
• Protection via user roles

Endress+Hauser

39Commissioning

Ecograph T, RSG35

Overview of access protection and security concept
Accessprotection

User

Setup
changes

Description

Open access-

Permitted

No protection, not recommended, all setup and system settings
are accessible.

Control input-

Permitted

Access protection via digital input (e.g. via key switch), all setup
and system settings are accessible if input actuated.

Access code-

Permitted

Access protection via access code, access authorization(distribution of access code) must be defined via (in-house)
regulations and securely controlled. All setup and systemsettings can be accessed after entering the access code.

Passwordprotected user
roles

Protection levels and access authorization can be defined via 3
levels of access (user roles). Access authorization (distribution ofpasswords) must be defined via (in-house) regulations andsecurely controlled.
Administrator Permitted

Access protection via administrator password, all setup andsystem settings can be accessed after entering the password.

Service

Permitted

Access protection via service password, all setup and systemsettings can be accessed after entering the password, advanced
functions (e.g. preset) are possible in the service mode.

Operator

LockedAll setup and system settings are locked, access to device
information and display values after entering the password.

In order to change any parameter, the correct code must first be entered or the device must
be unlocked using the control input.
Setup lock via control input: The settings for the control input can be found in the main
menu under "Setup -> Advanced setup -> Inputs -> Digital inputs -> Digital input X ->Function: Control input; Action: Lock setup".
It is preferable to lock the setup using a control input.
Setting up an access code: The settings for the access code can be found in the main
menu under "Setup -> Advanced setup -> System -> Security -> Protected by -> Accesscode". Factory setting: "Open access", i.e. modifications can always be made.
Make a note of the code and store in a safe place.
Setting up user roles: The settings for the user roles (operator, admin and service) areprovided in the main menu under "Setup -> Advanced setup -> System -> Security ->Protected by -> User roles" . Factory setting: "Open access", i.e. modifications can always
be made.
The passwords should be changed during commissioning.
Make a note of the code and store in a safe place.

9.9

HTTPS Web server setup

To operate the HTTPS Web server, an X.509 certificate and a suitable private key must be
installed on the device. For security reasons, installation is via a USB stick only.
The certificate that is preinstalled on the device when delivered from the factoryshould not be used!
Server certificates cannot be installed via the "USB stick/import SSL certificates"
function!

40

Endress+Hauser

Ecograph T, RSG35

Commissioning

Prerequisites
Private key:
• X.509 PEM file (Base64 encoded)• RSA key with max. 2048 bit• May not be password-protected
Certificate:• X.509 file (Base64 encoded PEM or binary DER format)• V3 incl. extension required• Signed by a certification authority (CA) or sub-certification authorities (recommended),
self-signed where necessary.
Certificate and private key can be created or converted using openssl(https://www.openssl.org) for example. Contact your IT administrator to create thecorresponding files.
Tip: More information on this topic is provided in our How To Videos underhttps://www.youtube.com/endresshauser
Installation:
1.

Copy the private key onto a USB stick into the root directory. File name: key.pem

2.

Copy the certificate onto a USB stick into the root directory. File name: cert.pem orcert.der3.

Connect the USB stick to the device. The private key and the certificate are installed
automatically. The installation is logged in the event logbook.

4.

Remove the USB stick using the "Safe removal" function
Notes:• The device may have to be restarted for the browser to use the new certificate• Delete the private key from the USB stick following installation• Keep the private key in a safe place• Use the private key and the certificate for one device only
• To prevent unauthorized use, it is possible to disable the USB A port on the device.
In this way, an attacker cannot replace the certificate or the private key ("Denial ofService"). If necessary, install a perimeter guard to prevent access to the device.

Checking certificates
You can check the certificate via "Main menu -> Diagnostics -> Device information ->SSL certificates". For this purpose, select the "Server certificate" point under thecertificate.
Replace the certificate in good time before it expires. The device will display adiagnostic message 14 days before the certificate expires.
Uninstalling certificates and the private key
You can check the certificate via "Main menu -> Diagnostics -> Device information ->SSL certificates". For this purpose, select the "Server certificate" point under thecertificate. You can delete the certificate here.
In this case, the preinstalled certificate is reused.
Using self-signed certificatesSelf-signed certificates must be stored in the PC's certificate memory under "Trusted
Root Certification Authorities" so that the browser does not display a warning.
Alternatively, an exception can be saved in the browser.

Endress+Hauser

41

Operation

Ecograph T, RSG35

10

Operation

The "Operation" menu is geared towards the tasks and activities of the operator. It contains
all the parameters that are needed in ongoing operation. Historical values and analyses,
for example, can be displayed in the "Operation" menu and display settings can be made.
Any settings made for the local display have no effect on the measurement section or theconfigured device parameters, however.
The device's simple operating concept and the integrated help function enables you to
perform operation for many applications without the need for hardcopy operating
instructions.

10.1

Reading-out and changing the Ethernet setting
address

To establish communication with the device via Ethernet, the following settings must be
known or modified where necessary:
Display IP/MAC address (only if DHCP is enabled): For the device's IP or MAC address,
see the main menu under "Diagnostics -> Device information -> Ethernet".
Display/change Ethernet settings: For the device's Ethernet settings, see the main menu
under "Setup -> Advanced setup -> Communication -> Ethernet".
Procedure to establish a direct connection via Ethernet (point-to-point connection):1.

Configure the PC (depends on operating system): e.g. IP address: 192.168.1.1; subnet
mask: 255.255.255.0; gateway: 192.168.1.1

2.

Disable DHCP on the device

3.

Make communication settings on the device: e.g. IP address: 192.168.1.2; subnet
mask: 255.255.255.0; gateway: 192.168.1.1
A crossover cable is not required.

10.2

Reading off the device locking status

If setup is locked via a control input, a padlock symbol appears on the top right of thescreen. The setup must first be unlocked via the control input before device parameters canbe edited.
Setup lock via control input: For the settings for the control input, see the main menu
under "Setup -> Advanced setup -> Inputs -> Digital inputs -> Digital input X ->Function: Control input; Action: Lock setup".
If setup is locked via the access code, all the operating parameters can be displayed, andcan also be edited as soon as the access code is entered.

42

Endress+Hauser

Ecograph T, RSG35

Operation

10.3

Reading off measured values (display devices)
11

10

9

12
8

7

6 5 4

3

2

1
A0047011

8
1
2
3
4
5
6
7
8
9
10
11
12

Front of device with open flap
Navigator: press briefly to open the main menu and confirm messages (=Enter); press for longer to open the
online help
Orange LED for read/write access to the SD card
Slot for SD card
USB B socket "Function"
Green LED lit: power supply present
USB A socket "Host"
Status bar
Area for measured value display (e.g. curve display)
Header: group name, analysis type
Header: current date/time
Header: alternating display indicating the percentage space on the SD card or USB stick that has already beenused. The display alternates between displaying memory information and status symbols.
Display of current measured values and the status in the event of an error/alarm condition. In the case ofcounters, the type of counter is displayed as a symbol.

An overview of all the symbols and icons is provided in the "Operation options" section.
→  28
If a measuring point has limit value status, the corresponding channel identifier ishighlighted in red (quick detection of limit value violations). During a limit value
violation and device operation, the acquisition of measured values continues
uninterrupted.
Information on how to rectify a problem if an error occurs is provided in the"Troubleshooting" section. →  58

10.4

Web server

The device features an integrated Web server, which enables access via Ethernet (orEthernet over USB). The Web server is used for convenient device commissioning andconfiguration and to visualize measured values. Access is possible from any access point
when the device is connected to an Ethernet network. An appropriate IT infrastructure,
Endress+Hauser

43

Operation

Ecograph T, RSG35

security measures etc. must be implemented in accordance with the requirements of theplant. Point-to-point access via Web server and Ethernet via USB is particularly well-suited
for service purposes.
Activation of the Web server in the menu Setup → Advanced setup → Communication →Ethernet → Web server → Yes or menu Expert → Communication → Ethernet → Webserver → Yes
The Web server port is preset to 80. The port can be changed in the Expert →Communication → Ethernet menu.
If the network is protected by a firewall, the port may need to be activated.
The following web browsers are supported:• MS Internet Explorer 11 and higher• MS Edge• Mozilla Firefox 52.1.0 and higher• Opera 12.x and higher• Google Chrome 66 and higher
A minimum resolution of 1920x1080 (full HD) is recommended.
In order to use the full functionality of the Web server, it is recommended that thelatest version of the browser is used. To access the device via a Web server, you musthave Administrator, Service or Operator authentication →  39
The Web server is not optimized for visualization on smartphones.
When delivered, the following default values are set for the user roles in the device andWeb server:• ID: admin; password: admin• ID: service; password: service• ID: operator; password: operator
Note: Passwords should be changed during commissioning!
The ID and password can be changed in the main menu under "Setup -> Advanced
setup -> Communication -> Ethernet -> Configuration Web server ->Authentication".

10.4.1

Access to the Web server via HTTP (HTML)When using an Internet browser, it suffices to enter the address http://<IP address> todisplay the information as HTML in the browser.
Note: Leading zeros in IP addresses must not be entered (e.g. enter 192.168.1.11
instead of 192.168.001.011).
As in the case of the display, you can alternate between the display groups in the Web
server. The measured values are automatically updated. In addition to the measured
values, status and limit value flags are displayed.

10.4.2

Access to the Web server via XML

XML format is available in addition to HTML format and contains all measured values of agroup. This can be integrated into additional systems as the user wishes.
The XML file is available in ISO-8859-1 (Latin-1) coding at http://<IP address>/values.xml (alternative: http://<IP address>/xml). However, some special characters,

44

Endress+Hauser

Ecograph T, RSG35

Operationsuch as the Euro symbol, cannot be displayed in this file. Texts such as digital statuses are
not transmitted.
Note: Leading zeros in IP addresses must not be entered (e.g. enter 192.168.1.11
instead of 192.168.001.011).
The decimal point is always displayed as a period in the XML file. All times are given
in UTC. The time difference in minutes is noted in the following entry.
The structure of the channel values for the XML file is explained as follows:<device

id="AI01IV" tag="Channel 1" type="INTRN"><v1>50.0</v1><u1>%</u1><vtime>20130506-140903</vtime><vstslvl1>0</vstslvl1><hlsts1>L</hlsts1><param><min>0.0</min><max>100.0</max><hh></hh><hi></hi><lo></lo><ll></ll></param><tag>Channel 1</tag><man>Manufacturer</man></device>
Tag

Description

device id

Unique ID of measuring point

tag

Channel ident.

type

Data type (INTRN, MODBUS)

v1

Measured value of channel as a decimal value

u1

Unit of measured value

vtime

Date and time

vstslvl1

Error level0 = OK, 1 = warning, 2 = error

hlsts1

Limit value status
H = upper limit value, L = lower limit value, LH = upper and lower limit value violation

param
min
max
hh
hi
lo
ll

Parameter (optional)
Lower zoom
Upper zoom
Upper alarm limit
Upper warning limit
Lower warning limit
Lower alarm limit

MAN

Manufacturer

10.4.3

Setup, operation and service via the Web server

Establish a connection to the Web server:
1.

Connect the PC to the device via Ethernet (or Ethernet over USB)

2.

Start the browser on the PC

3.

Enter the device's IP address in the browser http://<ip address>

4.

Log in with the ID and password

The startup screen of the Web server appears.

Endress+Hauser

45

Operation

Ecograph T, RSG35

Data Manager
DIN rail unit

A0037114

General information regarding the Device name, Device tag and Status signal isdisplayed in the top section of the Web server. The following functions can be accessed in
the middle of the screen:
Health status – Measured values – Menu – Data management - Network.
Clicking the functions calls up the following submenus. The submenus are closed by
selecting "Cancel" or by clicking "Back" several times.
Health status (advanced device status)
Function

Description

Device information Advanced device status, Ethernet settings, hardware configuration, device options,memory information, SSL certificate
Confirm messages

Submenu to confirm system messages

Event logbook

Submenu for the event logbook

Measured values (choice of measured value display)
Function

Description

Instantaneous values

Measured values are displayed as numerical instantaneous values, the signalgroups can be selected as tabs here

Instantaneous values ascurvesThe measured values are displayed as curves

History

Displays a history of the measured values

Menu
The menu structure displayed in the Web server corresponds largely to the menu structurein the device
Data management
Update functions and parameters relating to the firmware, load/save setup, save setup as
RTF, import SSL certificates
Network
Displays the Ethernet parameters (IP address, subnet mask, gateway, domain)46

Endress+Hauser

Ecograph T, RSG35

Operation

Displays the current measured values and history data
Under Measured values a selection window appears for Instantaneous values –Instantaneous values as curves – History. Click to display the corresponding display
function.
Current measured values can also be called up via the Web server without logging inwith the address http://<IP address>/iv. Device configuration is not possible in thiscase, however.
Note: The browser must support HTML5.
Optional command parameters:• Syntax: http://<ip address>/iv?group=<x>&refresh=<y>• group=<x> where x = 1 to 4
• refresh=<y> where y = 3 to 3600 in seconds
Note: Pay attention to lower and upper case if using an optional parameter.
This function can be disabled in the setup. If this function is disabled, the option toexport instantaneous values to XML is also disabled for security reasons.
Instantaneous values
The current measured values are displayed in numerical format. Clicking the tabs displays
the signal groups defined in the setup.

A0037118

Instantaneous values as curves
The current measured values are displayed as a curve and numerically over the time axis.
The refresh rate can be set in a selection window. The display mode can be set to fullscreen. Clicking the tabs displays the signal groups defined in the setup.

A0037117

Display functions
If the cursor is moved over one of the curves, the instantaneous value of the current pointin the curve is displayed with the time stamp and unit. Channels can be displayed andhidden using the channel checkboxes in the legend.

Endress+Hauser

47

Operation

Ecograph T, RSG35

History (logged measured values)
Clicking the History button loads the data previously logged. This may take a few seconds
depending on the data connection (USB, Ethernet, WLAN) and the number of measuring
channels. The data for each channel is loaded to display the content of one screen. The
logged (history) values are displayed as a curve and numerically over the time axis. The
display mode can be set to full screen. Clicking the tabs displays the signal groups definedin the setup.

A0037115

Display functions
If the cursor is moved over one of the curves, the instantaneous value of the current pointin the curve is displayed with the time stamp and unit. Channels can be displayed andhidden using the channel checkboxes in the legend.
Moving the dashed cursor line along the time axis updates the numerical display of themeasured values accordingly (on right).
Go to …: Enter a point in time. The history is reloaded. If you enter a time for which nomeasured values are available, the next possible time where measured values are available
is displayed.
Search older: The information displayed moves half a screen to the left (displays older
measured values). Press the << button to move the information one entire screen to theleft (older measured values).
Search newer: The information displayed moves half a screen to the right (displays more
recent measured values). Press the >> button to shift the information one entire screen to
the right (more recent measured values).
Note: If no measured values were recorded for a certain range of time (e.g. as there
was no power supply), this is indicated by a symbol at the top of the window. The
curves jump accordingly.

48

Endress+Hauser

Ecograph T, RSG35

Operation

A0037116

10.4.4

Remote control via the Web server

The device can be remote-controlled via the Web server. In the Web server, the remotecontrol function can be found under "Measured values -> Remote control". Here thedisplay shown corresponds directly to the device display. The device is operated using
buttons below this display. The interval for refreshing the display can be set in the"Refresh" menu.
Enabling remote control on the device:1.

In the menu Setup -> Advanced setup -> Communication -> Ethernet ->Configuration Web server -> Remote control, select "Yes" or under2.

Expert -> Communication -> Ethernet -> Configuration Web server -> Remotecontrol, select "Yes".

10.5

Changing the group

The group to be displayed can be changed in the main menu under "Operation -> Changegroup". Alternatively, the group can also be changed by turning the navigator.
Only the active groups are displayed here. The settings for this can be made in themain menu under "Setup -> Advanced setup -> Application -> Signal groups ->Group x".

10.6

Locking operation

Local operation can be blocked in the main menu under "Operation -> Lock operation" toprevent unintended or incorrect operation (e.g. when cleaning the device).
The device is unlocked by pressing the navigator or the OK operating key for 3 s.
When using an external keyboard, the device is unlocked with the key combination"Ctrl-Alt-Del".

Endress+Hauser

49

Operation

Ecograph T, RSG35

10.7

Login / logout

Log onto the device or log off the user currently logged on.
Only for role-based access protection →  39

10.8

SD card/USB stick

10.8.1

Function of SD card or USB stick

Without affecting the internal memory, data packets are copied block by block (min. 1 xdaily, midnight) to the SD card. Tests are also made to determine whether the data havebeen written without any errors. Once a new SD card is inserted, the device starts saving
the data automatically after 5 minutes. The use of a USB stick is only recommended if
certain data ranges should be copied.
The USB stick is not used for the continuous storage of measured values, i.e. it is not
updated automatically.
Data are saved in two different folders on the storage media depending on the storage
method:• All data are copied cyclically to the rec_data_<Device name> folder if a data packet iscomplete or the function "Update" is activated under "Operation -> SD card/USB stick > Update".
• The data for the selected time range that was selected under "Operation -> SD card /USB stick -> Save measured value" are copied to the rng_data_<device name> folder.
Copying these data does not affect the storage of the data in the rec_data_<device
name> folder.
• Only use new, formatted SD cards that are recommended by the manufacturer (see"Accessories" →  70).
• During normal operation, the used memory space on the SD card or USB stick isdisplayed on the top right of the display ("SD: xx%" or "USB: xx%").
Dashes "–" on this display mean that no SD card is inserted.
• The SD card must not be write-protected.
• Prior to removing the external data carrier, select "Operation -> SD card/USB stick-> Update". The current data block is closed and saved to the external data carrier.
This ensures that all the current data (up until the last save) is included on the datacarrier.
• Depending on the device configuration (see "Setup –> Advanced setup –> System–> Ext. memory –> Warning at"), an acknowledgeable message on the display
notifies the user that the external data carrier has to be changed before the datacarrier is 100 % full.
• The device keeps track of which data have already been copied onto the SD card or
USB stick. If the data carrier is not changed in time or if no SD card is inserted, the
new external data carrier is filled with the missing data from the internal memory insofar as the data are still in the memory. As measured value acquisition/recording
is assigned the highest priority, it may take several minutes in this case for the data
to be copied from the internal memory to the SD card or USB stick.

10.8.2Functions relating to the SD card or USB stick

For functions to save measured data and device settings on a removable medium, see themain menu under "Operation -> SD card / USB stick" (only if an SD card or USB stick isprovided).
Remove safely:
All internal access is terminated to ensure safe removal of the storage medium from thedevice. A message is displayed when the storage media can be removed. If the SD card is50

Endress+Hauser

Ecograph T, RSG35

Operation

not removed, the device automatically starts to save data to the storage medium again
after 5 minutes.
Only remove the data storage medium using this function; otherwise data may be
lost!
Update:Measurement data not yet saved on the storage medium are now saved. This may take amoment. Measured value acquisition is running in parallel and has top priority.
Data from several devices can be saved onto one storage medium.
• Save measured values:
A user-definable time range can be saved on the data storage medium.
• Load setup:
Loads device settings (setup) from the storage medium onto the device.
• Save setup:
All device settings (setup) are saved onto the storage medium. They can be archived orused for other devices.
• Save setup as RTF:
Saves the setup on the storage medium in a readable format as an RTF file (rich text
format).
The RTF file can be opened and formatted using suitable word processing software (e.g.
MS Word), allowing printouts to be easily made.
• Screenshot:
Save the current measured value display as a bitmap on the SD card or USB stick.
• Update firmware:
Loads new firmware onto the device. Only visible if a firmware file is provided on the SD
card or USB stick.
Caution: The device will restart. Save the setup and measured values beforehand on
the SD card or USB stick.
Import SSL certificate:
Uploads an SSL certificate (X.509) to the device. Certificates are needed to establish an SSL
connection in order to send e-mails in an encrypted manner for example. Certificates areavailable from your network administrator or provider. The following are supported: DER,
CER and CRT (binary or Base64-encoded).
Only visible if an SSL certificate is provided on the SD card or USB stick.

10.8.3

Notes on e-mail encryption

In addition to sending unencrypted e-mails, it is also possible to send encrypted e-mails via
SSL (TLS). To do so, you can choose either of two ways:• By SMTPS: fully encrypted via port 465.
The complete connection runs over TLS. The port is 465 by default but this value can bechanged in the Setup.
• Via port 25 or 587 using STARTTLS.
With this method the device first establishes a plain SMTP connection via port 25 andcontinues this connection following agreement and switchover to encryption.
The required process can be selected as follows: "Setup -> Advanced setup -> Application-> E-mail -> Server requires SSL" or under "Expert -> Application -> E-mail -> Server
requires SSL".
TLS V1.0 (=SSL 3.1), V1.1 and V1.2 are supported. Older standards are not supported. The
encryption method is automatically agreed with the counterparty.

Endress+Hauser

51

Operation

Ecograph T, RSG35

A certificate must be installed in order to be able to send encrypted e-mails. Thesecertificates can be obtained from your e-mail service provider. The following file formats
are supported:• *.CER: DER- or Base64-encoded certificate• *.CRT: DER- or Base64-encoded certificate• *.DER: DER-encoded certificate
The file name of the certificate may only contain the following characters: a..z, A..Z,
0..9, +, -, _, #, (, ), !
To establish an SSL connection, the device automatically selects the certificate that best
suits the counterparty from all the installed certificates. An error message is displayed if
the device does not have any of the required certificates.
When e-mail encryption is enabled, no e-mails can be sent if a valid certificate is notavailable or if the certificate has expired.

10.8.4

Notes on WebDAV encryption

In addition to sending unencrypted data to the WebDAV server, it is also possible to send
encrypted data via SSL (TLS). All data are sent in encrypted format via the SSL port of theexternal WebDAV server.
The complete connection runs over TLS. The port is 80 by default but this value can bechanged in the Setup. The procedure to be used can be selected as follows: "Setup ->Advanced setup -> Application -> WebDAV Client -> Enable -> Yes (SSL)" or under"Expert -> Application -> WebDAV Client -> Enable -> Yes (SSL)".
TLS V1.0 (=SSL 3.1), V1.1 and V1.2 are supported. Older standards are not supported. The
encryption method is automatically agreed with the counterparty.
A certificate must be installed in order to be able to send encrypted data. These certificates
can be obtained from your WebDAV server service provider. The following file formats aresupported:• *.CER: DER- or Base64-encoded certificate• *.CRT: DER- or Base64-encoded certificate• *.DER: DER-encoded certificate
The file name of the certificate may only contain the following characters: a..z, A..Z,
0..9, +, -, _, #, (, ), !
To establish an SSL connection, the device automatically selects the certificate that best
suits the counterparty from all the installed certificates. An error message is displayed if
the device does not have any of the required certificates.
When WebDAV client encryption is enabled, no data can be sent if a valid certificate
is not available or if the certificate has expired.

10.8.5

SSL certificates

Importing an SSL certificate
Installing a certificate via SD card or USB stick:1.

Copy a certificate on a PC to an SD card or USB stick2.

Insert the SD card or USB stick into the device

3.

In the main menu, select "Operation -> SD card (or USB stick) -> Import SSL
certificate"4.

Select the required certificate from the list and follow the dialog on the display.

Installing a certificate via the Web server:
1.

52

On the Web server, select "Data management --> Import SSL certificate"Endress+Hauser

Ecograph T, RSG35

Operation2.

Select the file3.

Start the process by clicking OK
Up to 3 certificates can be installed simultaneously.

Verifying the installed SSL certificates
The installed certificates can be verified in the main menu under "Diagnostics -> Device
information -> SSL certificates". The most important certificate information, e.g. subject
key ID, organization and validity period, are displayed in the parameter list.
Not all fields are completed on all the certificates. This is because the parties issuing
the certificates do not make all the information available.
Deleting an SSL certificate
In the main menu select the certificate to be deleted under "Diagnostics -> Device
information -> SSL certificate -> Certificate" and select "Yes" under "Delete certificate".
Period of validity of certificates
Certificates are valid for a defined period (valid from ... to …). The device checks thecertificate validity once a day or each time the device is rebooted. 14 days before thecertificate expires the device notifies the user daily (by e-mail, screen display, entry in
event log) that the certificate will expire shortly.
If the certificate has expired the alarm relay switches (if activated) and a message isdisplayed on the screen. An entry is also made in the event log. If a certificate is deleted all
errors related to this certificate are reset.

10.9

Show data logging

You can scroll through the saved measured values in the main menu under "Operation ->History". Turn the navigator clockwise or counterclockwise to scroll back and forth
between the measured value curves. Press the navigator to make other settings for thehistorical data display (e.g. scroll speed, time scaling or change display mode) or to quit thehistorical data function.
The gray header on the screen and the symbol in the status bar indicate that
historical values are being displayed. In the instantaneous value display the color of
the header is blue.

10.9.1

Historical data: changing a group

It is possible to change the group to be displayed in the historical data under "Operation ->Change group" in the data history.

10.9.2

Historical data: Scroll speed

Specify how fast the display should scroll when the navigator is turned.
It is possible to change the scroll speed in the historical data under "Operation -> Scroll
speed" in the data history.
The scroll speed can also be set via the soft key with the arrow symbol < or >. The speed
can be changed from < (slow) to <<<< (fast) by pressing the soft key repeatedly.

Endress+Hauser

53

Operation

Ecograph T, RSG35

10.9.3

Historical data: Time scaling

It is possible to scale the displayed time range in the historical data under "Operation ->Time scaling" in the data history.
Notes:• "1:1" option: Every measured value is displayed.
• "1:n" option: Only every nth measured value is displayed (increases the displayed
time range).
• No interpolation is performed, nor is the mean value determined.
• If the value for "n" is large, this can result in extended loading times.
• Time scaling does not affect the process for storing the measured value.
• The time range displayed per screen for the currently set time scaling is also
displayed in the menu.

10.9.4

Historical data: Time range displayed

The displayed time range is shown in the historical data under "Operation -> Time range
displayed" in the data history. This tells the user which time range is displayed per screenin the standard memory cycle.
If the alarm cycle differs from the standard memory cycle, this is not taken into
consideration.

10.9.5

Historical data: Screenshot

The current measured value display can be saved as a bitmap to an SD card or USB stick in
the historical data under "Operation -> Screenshot".

10.9.6

Historical data: Change the display mode

It is possible to change the display mode of the active group in the historical data under"Operation -> Change display mode" in the data history.
The following display modes are possible: Curve, Curve in ranges, Waterfall, Waterfall in
ranges.
The various display modes have no influence on the signal recording.

10.10 Signal analysis
The analyses saved in the device are displayed in the main menu under "Operation ->Signal analysis".
• Actual intermediate analysis:
The current (i.e. not yet completed) intermediate analysis can be displayed here.
• Actual day:
The current (i.e. not yet completed) daily analysis can be displayed here.
• Actual month:
The current (i.e. not yet completed) monthly analysis can be displayed here.
• Actual year:
The current (i.e. not yet completed) annual analysis can be displayed here.
• Search:
Search and display of analyses. Select which analyses the system should search for/display: Intermediate analysis, Daily analysis, Monthly analysis, Annual analysis.

54

Endress+Hauser

Ecograph T, RSG35

Operation

10.11 Search in trace
In the main menu, the internal memory can be searched for events or times under"Operation -> Search in trace".
Search for events: The event logbook forms the basis for searches for events. To searchmore easily for certain events (e.g. setup changes), the search filter can be used to selectand search for the desired events. In the standard all events are shown. In the event list
displayed, it is possible to select an event and go directly to this point in the history (if stillin the memory).
Search for time: When searching for a time in the past, the user can enter a date and atime to specify when the system should start displaying historical data. Once the date/time
have been entered and confirmed, the display goes to the selected time in the active group.

10.12 Changing the display mode
The display mode of the active group can be changed in the main menu under "Operation-> Change display mode".
The following display modes are possible: Curve, Curve in ranges, Waterfall, Waterfall in
ranges, Bargraph and Digital display.
The various display modes have no influence on the signal recording.

10.13 Adjusting the brightness of the display
You can adjust the brightness of the display in the main menu under "Operation -> Adjust
brightness":
Parameter

Possible settings

Description

Adjust brightness

0-100
Default: 80

Sets the brightness of the display

10.14 Limit values
It is possible to change the limit values during operation in the main menu under"Operation -> Limits".
This function must be enabled beforehand in the main menu under "Expert ->Application -> Limits -> Change limits: outside of setup also".
Detailed description of the limit values: →  158

10.15 WebDAV client
The function of the WebDAV client is to transmit recorded data automatically to aconnected WebDAV server (e.g. NAS drive). The recorded data are sent to the connected
WebDAV server cyclically every 15 minutes. The generated files correspond to the files
that are automatically saved to the SD card.
The client is configured via "Setup -> Advanced setup -> Application -> WebDAV
Client". The settings under "Setup -> Advanced setup -> System -> External memory"
are also used, with the exception of the settings for the SD card (Memory build-up,
Warning and Relays). The memory is regarded as a stack memory.
Detailed description of the parameters: →  175
Note: With the WebDAV client, the data are transmitted to the WebDAV server inaccordance with the selection made in the .CSV or "Protected format".
Endress+Hauser

55

Operation

Ecograph T, RSG35

10.15.1 Access to the WebDAV server via HTTP (HTML)
Address entry in browser: http://<ip-address>/webdav
The data are updated cyclically every 15 minutes. The data are automatically updated eachtime a user logs in.
Note: Leading zeros in IP addresses must not be entered (e.g. enter 192.168.1.11
instead of 192.168.001.011).
Administrator or service authentication is required. ID and password administration is
performed in the main menu under "Setup -> Advanced setup -> Communication ->Ethernet -> Configuration Web server -> Authentication".
ID default value: admin; Password: admin
Note: The password should be changed during commissioning!
Note: For devices with a stainless steel front and touchscreen, the data are always
made available in "Protected format" via the WebDAV server.

10.16 Data analysis and visualization with the Field DataManager software (FDM) provided
The analysis software offers centralized data management with visualization for recorded
data.
This enables the data of a measuring point to be fully archived, e.g.:• Measured values• Diagnostic events• Protocols
The analysis software stores data in an SQL database. The database can be operated locallyor in a network (client/server). The free PostgreTMSQL database supplied on the DVD canbe installed and used.
For details, see the Operating Instructions on the analysis software DVD provided.

10.16.1 Structure/layout of a CSV file
The CSV files are comprised as follows:
File name (=serial number + Description
file number + configuration
number + date and start
time + data type)

Coding

H4000504428 0000000279
0000000185 2013-11-07
11-18-00 GROUP01.csv

Contains all the measured values of the group from thestarting time indicated in the file name. An individual CSV
file is created for each group.

ANSI

H4000504428 0000000279
0000000185 2013-11-07
11-30-00 ANALYSIS01.csv

Contains the signal analyses of the active channels fromthe starting time indicated in the file name. An individual
CSV file is created for every analysis (01 - 04).

ANSI

H4000504428 0000000279
2013-11-07 11-18-34
EVENTS.csv

Contains the event log from the starting time indicated in
the file name.

Unicode UTF-8(see the notes in
the following
section)Meaning of the values under "Status" and "Limit" when analyzing a group:
Status of the channel:
0: OK
1: Cable open circuit

56

Endress+Hauser

Ecograph T, RSG35

Operation

Status of the channel:
2: Input signal too high3: Input signal too low
4: Invalid measured value
6: Error value, i.e. not the calculated value (for mathematics if an input variable is invalid)
7: Sensor/input error
Bit 8: Not assigned
Bit 9: Alarm storage
Bit 10..13: Not assigned
Bit 14: Use error value
Bit 15: Not assigned

General status:
1: High speed storage is active
2: Additional hour for summer/normal time changeover
Note: A combination of 1 and 2 is also possible.

Status of the limit value ("Limit"):
0: OK, no limit value has been violated
Bit 0: Lower limit value
Bit 1: Upper limit value
Bit 2: Increasing gradient
Bit 4: Decreasing gradientNote: A combination is also possible.

10.16.2 Importing UTF-8-encoded CSV files into spreadsheets
You might encounter problems displaying information if importing UTF-8-encoded CSV
files directly into more recent versions of MS ExcelTM (2007 and higher).
Importing CSV data from the event logbook ("Events") into MS ExcelTM (version 2007
and higher):Endress+Hauser

1.

In the menu select "Data -> Get external data - From text"

2.

MS Office 365 and higher: In the menu select "Data -> From text/CSV"

3.

Select the CSV file4.

Follow the instructions in the wizard

5.

Select file origin "Unicode UTF-8"

57

Diagnostics and troubleshooting

Ecograph T, RSG35

11

Diagnostics and troubleshooting

The following section contains an overview of possible causes of errors to provide initialassistance during troubleshooting.

11.1

General troubleshooting

LWARNING
Danger! Electric voltage!
‣ Do not operate the device in an open condition for error diagnosis!
Display

Cause

Remedial action

No measured value display; no
LED litNo supply voltage connected

Check the device supply voltage.

Supply voltage is applied; device orpower unit is defective

Replace the power unit or device.

Diagnostic message is displayed

For a list of diagnostic messages, see the next section.

Dead pixels: Dead pixels refer to pixels on LCD and TFT displays that are defect due tothe technology or manufacturing techniques used. The TFT display used can have up
to 10 dead pixels (Class III as per ISO 13406-2). These dead pixels do not entitle theuser to a warranty claim.

11.2

Troubleshooting

The Diagnostics menu is used for the analysis of the device functions and offers
comprehensive assistance during troubleshooting. Always proceed as follows to locate thecause of device errors or alarms.
General troubleshooting procedure
1.

Open diagnosis list: lists the 30 most recent diagnostic messages. This can be used todetermine which errors are currently present and whether an error has repeatedly
occurred.

2.

Diagnosis of current measured values: Verify the input signals by displaying thecurrent or scaled measuring ranges. To verify calculations, call up calculated auxiliary
variables if necessary.

3.

Most errors can be rectified by performing steps 1 and 2. If the error persists follow
the troubleshooting instructions in the following sections.

4.

If this does not rectify the problem, contact the Service Department. Any time youcontact the Service Department please always have the error number and theinformation in the main menu under "Diagnostics -> Device information" (program
name, serial number etc.) to hand.

The contact data for your Endress+Hauser representative can be found on the Internet atwww.endress.com/worldwide.

11.2.1

Device error/alarm relay

One relay can be used as an alarm relay. If the device detects a system error (e.g. hardware
defect) or a malfunction (e.g. cable open circuit), the selected output/relay switches. The
alarm relay is assigned in the main menu under "Setup -> Advanced setup -> System ->Fault switching -> Relay x". Factory setting: Relay 1.
This "alarm relay" switches if "F"-type or "S"-type errors occur, i.e. "M"-type or "C"-type errors
do not switch the alarm relay.

58

Endress+Hauser

Ecograph T, RSG35

Diagnostics and troubleshooting

11.3

Diagnostic information on the local display

The diagnostic message consists of a diagnostic code and a message text.
The diagnostic code is made up of the error category as per Namur NE 107 and themessage number.
Error category (letter in front of the message number)• F = Failure. A malfunction has been detected.
The measured value of the affected channel is no longer reliable. The cause of themalfunction is to be found in the measuring point. If a controller is connected, it should
be switched to manual mode. An alarm relay can be assigned to this error category in
the Advanced setup.
• M = Maintenance required. Action must be taken as soon as possible.
The device still measures correctly. Immediate measures are not necessary. Proper
maintenance efforts may prevent a possible malfunction in the future.
• S = Out of specification. The measuring point is being operated outside specifications.
Operation is still possible. There is the risk of increased wear, a shorter operating life orless accurate measurements. The cause of the problem is to be found outside themeasuring point.
• C = Function check. The device is in Service mode.
Diagnosticcode

Message text

Description

RemedyF100

Sensor/input error!

Sensor/input error!

Check connections and parameters

F101

Open circuit

Open circuit

Check connections

F105

Invalid value!

Measured value is invalid (when calculating -->NAN)

Check connections and process variables

F201

Device fault

Device error

Contact the Service Department

F261

Error: RAM

No access to RAM

Contact the Service Department

F261

Error: FlashNo access to flash

Contact the Service Department

F261

Error: SRAM

No access to SRAM

Contact the Service Department

F261

Analog card x is out oforder!

Hardware defect detected

Contact the Service Department, replace card

F261

Power supply out of order!

Hardware defect detected

Contact the Service Department, replace power unit

M284

Firmware update

Firmware has been updated

No action required. Message can be acknowledged.

M290

Internal flash reaches theend of its lifetime. Please
replace device.

The internal flash memory is defective.

Replace device

F301

Error: Cannot load setup

Setup defective

Switch the device off and then on again, re-configure,contact the Service Department if necessary

M302

Setup restored from backup

Setup has been loaded from backup

Check setup

F303

Error: Device data

Device data defective

Contact the Service Department

M304

Backup: Device data

Device data defective. However it was possible
to continue working with the backup data.

Check settings (e.g. serial number)

F307

Error: Customer preset
value defective

Customer preset value defective

F309

Error: Date/time is not set

Invalid date/time (e.g. internal battery is empty) Device was switched off too long. The date/time must
be set again. Battery might need to be replaced(contact the Service Department).

F310

Error: Cannot save setup

The setup could not be saved

Contact the Service Department

F311

Error: Device data

The device data could not be saved

Contact the Service Department

F312

Error: Calibration data
defective

The calibration data could not be saved

Contact the Service Department

Endress+Hauser

59Diagnostics and troubleshooting

Ecograph T, RSG35

Diagnosticcode

Message text

Description

RemedyF312

Analog card x is notcalibrated!

Analog card x is not calibrated! The device
works with default values, i.e. the measured
values may be inaccurate under certain
circumstances.

Contact the Service Department

M313

SRAM has been
defragmented

SRAM was defragmented after firmware update No action required. Message can be acknowledged.

F314

Error: Option code

Activation code is no longer correct (incorrect
serial number/program name). Option was
disabled and setup preset has been performed.

Enter new code

M315

No IP address could beobtained from the DHCP
server!

No IP address could be obtained from the DHCP
server!

Check the network cable

M316

Invalid MAC address!

No or incorrect MAC address

Contact the Service Department

M317

Battery voltage < 2.5 V.
Please replace battery!

F348

Firmware cannot be
updated:• Checksum incorrect• Firmware incompatible!

Firmware update has been aborted because thefirmware file is damaged or is not compatible
with this device

Contact the Service Department

M350

Measured value acquisitioninterrupted for calibration/
service work.

Measured value acquisition was interrupted/
reactivated for service/maintenance purposes.

No action required. Message can be acknowledged.

M351

Battery needs to be replaced (contact the Service
Department)

Measured value acquisition
restarted.

Causes include:• Calibration of inputs/outputs• Firmware update

The device is restarted.

The device is rebooting.

No action required. Message can be acknowledged.

Causes include:• Following a firmware update• Change to device options
F431

Error:Calibration

Calibration data missing

Contact the Service Department

M502

Device is locked!

Device is locked! The message appears when an
attempt is made to update the firmware, forexampleCheck lock per digital channel

F510

Setup was corrected.

The device has discovered that theconfiguration is no longer correct. All theparameters concerned have been reset to thefactory default settings.

Check the configuration of the device. If hardware has
been replaced, no other action is needed(recommendation: change the operating language sothat the error message no longer appears after the
next restart).

Possible causes:• Input cards have been removed or replaced byanother type
• An input card is no longer working correctly
A firmware update has caused compatibility
problems.
Note: This error message appears each time thedevice is restarted until at least one change has
been made to the configuration.
M520

SMTP: Name could not be
resolved (DNS)!

Problem with name resolution (DNS).

SNTP: Name could not be
resolved (DNS)!

SNTP: time synchronization

Check the corresponding settings

SMTP: e-mailM528

Setup is not compatible with An attempt was made to load a setup which is
this firmware
not compatible with this firmware (e.g. anotherdevice type)

Check whether the correct file has been selected.

M530

Cannot copy setup.

Replace the SD card or USB stickAn error occurred when a setup was loaded
from an SD card or USB stick

Setup file defective?

An error occurred when a setup was saved to an
SD card or USB stick60

Endress+Hauser

Ecograph T, RSG35

Diagnostics and troubleshooting

Diagnosticcode

Message text

Description

RemedyS901

Input signal too small

Input signal too small

Check connections and parameters. Check connected
sensor/transmitter.

S902

Input signal too highInput signal too highCheck connections and parameters. Check connected
sensor/transmitter.

M905

Limit xLimit x has been violated

Note: Error number only occurs if e-mails are sent

M906

End limit x

Limit x no longer violated

Note: Error number only occurs if e-mails are sent

F910

This software is not enabled The current firmware is not enabled for thisfor this device.
hardwareContact the Service Department

M920

Too many messages thatneed to be acknowledged!

There are too many messages that need to beacknowledged. Another message could not be
added.

Acknowledge messages

M921

SD card x% full.

External memory is full

Replace SD card

M922

No cyclic value readout

The instantaneous values were not read out fora set time

M922

No cyclic transfer

The device was not read out via fieldbus for a
configurable time

Check the communication of the fieldbus. Check PLC.

M924

Error accessing SD card!
Error accessing USB stick!
SD card is not or wrong
formatted!
USB stick is not or wrong
formatted!

Impossible to access the removable data
medium. Causes include:
Memory is larger than 32 GB

Check/replace removable data medium

M925

SD card is write-protected!

SD card is write-protected!

Remove write protection

M927

Insufficient space free ondata storage medium!

An attempt was made to save to the SD card or
USB stick (setup, screenshot, etc.), but notenough free memory space is available.

Use other SD card/USB stick. Delete files that are nolonger needed from the SD card/USB stick

M927

Insufficient space free ondata storage medium!

An attempt was made to save to the WebDAV
server but not enough free memory space isavailable.

Use other WebDAV server. Delete files that are nolonger needed from the WebDAV server.

F929

File is damaged!

The file that should be loaded is damaged/
invalid (e.g. wrong checksum).

Create file again, use other storage medium.

Invalid format (only FAT or FAT32 arepermitted)

This message can occur in connection with thefollowing actions, for instance:• Loading setup from SD card/USB stick• Firmware update
M940

M941

E-mail could not be sent! (x) E-mail could not be sent!

No connection to the e-mailserver!

Endress+Hauser

Check settings/network connection

Optional: Error code (x) from server: e.g.:• 451: Requested action aborted: local
processing error• 554: Transaction failed. Possible reason: email was not sent as suspected of being
SPAM• 1: No free buffer• 2: No receiver specified• 451: Try again• 554: Use other e-mail provider

A connection to the e-mail server could not beestablished because:• The entered connection data are incorrect• The connection is lost

Check settings/network connection

61

Diagnostics and troubleshooting

Ecograph T, RSG35

Diagnosticcode

Message text

Description

RemedyM942

SMTP: Fault occurred (x).

An error occurred when sending an e-mail.
x= error code:
0: SMTP was switched off when the mail was
being sent
3: TCP/IP connection was denied
4: TCP/IP connection error
5: SMTP server denied
6: Error during authentication
7: Connection unexpectedly lost
8: Server responded with error code
9: Timeout
10: Internal protocol error

Check settings/network connection

M944

SMTP: authentication
failed!

M945

SNTP: Time was notsynchronized!

Time could not be synchronized via SNTP.

SNTP server 1 not
responding. Try server 2.

Time could not be synchronized via SNTP.

M946

Screenshot could not besaved (x)!

Screenshot could not be created.
Possible causes (x):0: Error when writing
1: Insufficient free space
2: Bitmap could not be created
3: SD card/USB stick not available or not yet
ready

Check/replace the SD card or USB stick

M947

Modem could not be
initialized! Check the cableand modem.

The connected modem could not be initialized
by the device.

Check the cable and modem.

M950

Cannot load SSL certificate.

Cannot load SSL certificate. Cause:• Use a certificate with a valid file format• Import certificate to device again

M945

Check settings/network connection

Possible reasons:• SNTP server temporarily unavailable• Settings not correct

Possible reasons:• SNTP server temporarily unavailable• Settings not correct• Invalid file format• File is damaged• Check settings• Check whether the error occurs often. If it does,choose another time server.

• Check settings• Check whether the error occurs often. If it does,choose another time server.

F951

SSL certificate '…' has
expired!

Certificates have an expiry date, i.e. they must
be renewed from time to time.

Install a new certificate

M952

SSL certificate '…' expireson ...!

The device warns the user shortly before thecertificate expires.

Install a new certificate

M953

x certificates have alreadybeen installed. Delete
certificates that are nolonger required.

The device can manage a maximum of 3 X.509
certificates.

Delete a certificate that is already installed and nolonger required

M954

SSL certificate not found:
key ID = …

Unable to establish an SSL connection as asuitable certificate is not installed.

Install a suitable certificate

M955

SSL connection denied!

M980

No connection to the
WebDAV server

A connection to the WebDAV server could not
be established because the entered connection
data are incorrect or the connection has beeninterrupted.

Check settings/network connection

M981

WebDAV: authentication
failed!

M982

WebDAV: directory or file
could not be created!

Configured directory path not available.

M983

WebDAV: Fault

An unassigned error has occurred. The error isdisplayed in English.

62

Check settings
Create a directory manually in the WebDAV serverEndress+Hauser

Ecograph T, RSG35

Diagnostics and troubleshooting

Diagnosticcode

Message text

Description

RemedyM984

No Ethernet connection.

The device is not connected by an Ethernet
cable.

Establish cable connection.

M985

The test cannot be carriedout because data are
currently being copied by
WebDAV.

M988

Server certificate cannot beloaded. Invalid format.

The file must be Base64-encoded.
Format: X.509 certificate V3 incl. extension

Create the certificate again according to thespecifications

M989

Private key cannot beloaded. Invalid size/format.

The file must be Base64-encoded. Only RSA
keys with max. 2048 bits are supported.

Create the certificate again according to thespecifications

M990

Server certificate cannot be
installed.

General error. Unable to read or write to file.

Check the files on the USB stick and create again if
necessary. If the error persists, contact the Service
Department.

Repeat later

11.4

Pending, current diagnostic messages

The diagnostic message that is currently active, the last diagnostic message and the lastdevice restart are displayed in the main menu under "Diagnostics -> Actual diagnostics","Diagnostics -> Last diagnostics" or under "Diagnostics -> Last restart".

11.5

Diagnosis list

The last 30 diagnostic messages are displayed in the main menu under "Diagnostics ->Diagnosis list" (messages with Fxxx, Sxxx or Mxxx-type error numbers).
The diagnosis list is designed as a ring memory, i.e. when the memory is full the oldest
messages are automatically overwritten (no message).
The following information is saved:• Error number• Error text
• Date/time

11.6

Event logbook

Events such as alarm set point infringement and power failure are displayed in the correcttime sequence. It can be found in the main menu under "Diagnostics -> Event logbook".
Individual events can be selected and details displayed.

11.7

Device information

Important device information, e.g. serial number, firmware version, device name, device
options, memory information, SSL certificates etc. are displayed in the main menu under"Diagnostics -> Device information" .
For further information open the online help on the device.

11.8

Diagnostics of measured values

Displays the current measured values in the main menu under "Diagnostics -> Measured
values". The input signals can be verified here by displaying the scaled and calculated
values. To verify calculations, call up calculated auxiliary variables if necessary.

Endress+Hauser

63Diagnostics and troubleshooting

Ecograph T, RSG35

11.9

Diagnostics of outputs

Displays the current states of the outputs (relays 1-6) in the main menu under"Diagnostics -> Outputs".

11.10 Simulation
Various functions/signals can be simulated for test purposes here.
NOTICE
Invoke simulation: for the simulation of the relays, see the main menu under"Diagnostics -> Simulation". For the simulation of the measured values, see the main
menu under "Expert -> Diagnostics -> Simulation".
Only the simulated values are recorded during simulation. The simulation is recorded in
the event logbook.
‣ Do not start simulation if measured value recording must not be interrupted!

11.10.1 E-mail test
A test mail can be sent to the selected recipient in the main menu under "Diagnostics ->Simulation -> E-mail".
At least one e-mail address must be set beforehand.

11.10.2 Test WebDAV client
A test file can be sent to the selected WebDAV server in the main menu under"Diagnostics -> Simulation -> WebDAV Client".
The settings for the WebDAV server to be addressed must be set beforehand under"Setup -> Advanced setup -> Application -> WebDAV Client".

11.10.3 Test time synchronization/SNTP
Time synchronization (SNTP setting) can be tested in the main menu under "Diagnostics > Simulation -> SNTP".
SNTP must be enabled beforehand in the main menu under "Setup -> Advanced
setup -> System -> Date/time set-up -> SNTP".
Note: The test can take some time.

11.10.4 Relay test
The relay selected under "Diagnostics -> Simulation -> Relay x" can be switchedmanually in the main menu.

11.11 Initializing the modem
Initializes the modem connected (to automatically answer calls). The modem must support
the complete AT command syntax.
• Set the baud rate in the main menu under "Setup -> Advanced setup ->Communication -> Serial interface" and select "RS232" as the interface type.
• Connect the modem to the RS232 interface of the device. For this purpose only use
the modem cable, which is available as an accessory.
A GSM modem can only be initialized if a SIM card is inserted and the PIN is enteredor the prompt to enter the PIN has been disabled.

64

Endress+Hauser

Ecograph T, RSG35

Diagnostics and troubleshooting

11.12 Resetting the measuring device
The device can be reset to the as-delivered state with a PRESET. This function should only
be performed by a service technician.
The function can be found in the main menu under "Expert -> System -> PRESET"PRESET is only visible under "Expert" once the service code has been entered.
Procedure for resetting the measuring device
The PRESET returns all parameters to the factory default setup! The internal memory
content is deleted!

‣ Save the setup and measured values on the USB stick or SD card. Then perform a
PRESET.
 The device is reset to the factory default settings.

11.13 Clear memory
The internal memory should be cleared after commissioning so that the analysis
software does not contain any unnecessary data.

11.14 Reset analysis
The analyses should be reset after commissioning so that the analysis software does
not contain any unnecessary data.

11.15 Firmware history
Overview of unit software history:

Endress+Hauser

Unit software
Version / date

Software changes

FDM analysis
software version

Version of OPC server

OperatingInstructions

V01.00.00 /07.2013

Original software

V01.01.02.10 andhigher

V5.00.02.04 and higher

BA01146R/09/01.
13

V01.01.00 /02.2014E-mail via SSL;
V01.02.00.08 andadded functionality higher

V5.00.02.04 and higher

BA01146R/09/02.
14

V02.00.00 /08.2015

Added
functionality

V01.03.00.00 andhigher

V5.00.03.00 and higher

BA01146R/09/03.
15

V2.01.00 /04.2016

Extendedfunctionality/bug
fixes

V01.03.01.00 andhigher

V5.00.03.00 and higher

BA01146R/09/04.
16

V2.01.05 /11.2016

Extendedfunctionality/bug
fixes

V01.03.01.01 andhigher

V5.00.03.00 and higher

BA01146R/09/05.
16

V2.02.00 /11.2017

Ethernet function
via USB

V1.04.00 andhigher

V5.00.04.00 and higher

BA01146R/09/06.
17

V2.04.00 /09.2018

Web server
extension

V1.04.02 andhigher

V5.00.04.01 and higher

BA01146R/09/07.
18

V2.04.05 /08.2021

Support for HTTPS
server; bugfixes

V1.04.02 andhigher

V5.00.04.01 and higher

BA01146R/09/08.
21

65

Maintenance

Ecograph T, RSG35

12MaintenanceNo special maintenance work is required for the device.

12.1

Updating the device software ("firmware")Updating the device software ("firmware") via USB stick, SD card or Web server.
The function to update the firmware via the Web server must be enabled beforehand
under "Expert -> Communication -> Ethernet -> Configuration Web server".
There are two ways to update the firmware:• In the main menu under "Operation -> SD card or USB stick -> Update firmware"• In the Web server under "Data management -> Update firmware"
It is advisable to save the setup and measured values beforehand on the SD card or
USB stick.
The device software ("firmware") should only be updated by a service technician.
The device will restart following the firmware update.
If an older firmware version (< V2.04.00) is installed on the device, the internal
memory must then be cleared under "Expert -> System".

12.2

Instructions for enabling a software option

Various device options can be enabled via an activation code. Available device options canbe ordered as an accessory →  70. Once you place your order, you will receive
instructions on how to activate the option and a code which must be entered under "Mainmenu -> Expert -> System -> Device options -> Activation code".

12.3

Cleaning

The front of the housing can be cleaned with a clean dry or damp cloth.

66

Endress+Hauser

Ecograph T, RSG35

Repair13

Repair

13.1

General notes

Repairs that are not described in these Operating Instructions must only be carried out
directly by the manufacturer or by the service department.
Always quote the serial number of the device when ordering spare parts! Installation
instructions are included with the spare part.

13.2

Spare parts

Information on accessories and spare parts that are currently available for the productcan be found online at: www.endress.com/spareparts_consumables → access to
specific device information → enter serial number.

17
18

19
20
1

16
15
14

2

3
4
5

13

10
12

9
8
7

11

6

A0019863

9

Spare parts diagram

Spare parts list:

Endress+Hauser

Item No.

Description

Order No.

1

Casing71155332

12

Housing fastening unit, short (1 pc)

71035184

11

Housing seal

71155329

16

Analog card (4 channels)

XPR0007-A1

4

Display TFT 5.7" VGA + flat ribbon connector

XPR0007-A2

3, 5, 10

Spare parts kit, display

XPR0007-A3

6, 7, 8Front + navigator + flat ribbon connector

XPR0007-A4

6, 7, 8Front neutral + navigator + ribbon cable connector

XPR0007-A5

15, 20

Spare parts kit, card holder

XPR0007-A6

67

Repair

Ecograph T, RSG35

Item No.

Description

Order No.

14

Motherboard

XPR0007-B1

18

Power supply 24 V AC/DC

XPR0007-B2

18

Power supply 100-230 V AC (+/-10%)

XPR0007-B3

17

Rear panel analog inputs

71165643

13

Terminals:
Plug-in terminal 3-pin for mains connection "N L PE" RM5.08 –orange color

71123475

Plug-in terminal 3-pin FKC2.5/3-ST-5.08 for relay 1 (changeover)

71037408

Plug-in terminal 4-pin FKC2.5/4-ST-5.08 for relay 2+3

71037410

Plug-in terminal 6-pin FKC2.5/6-ST-5.08 for relay 4+5+671037411

Plug-in terminal 9-pin FMC1.5/9-ST-3.5 for digital inputs

71037363

Plug-in terminal 6-pin FMC1.5/6-ST-3.5 for analog input

51009211

Product structure for CPU with software
Item No.

Description

Order code19

CPU + software

XPR0008-_ _ _ _Operating language:
Universal

XPR0008-A1

Software:Standard
Mathematics

XPR0008-A1A
XPR0008-A1B

Communication:
Ethernet RJ45 + USB
RS232/485 + Ethernet RJ45 + USB
Modbus TCP slave + Ethernet RJ45 + USB
Modbus RTU/TCP slave + RS232/485 + Ethernet RJ45 + USB

XPR0008-A1_ A
XPR0008-A1_ B
XPR0008-A1_ C
XPR0008-A1_ D

Option:Standard
Neutral

XPR0008-A1_ _ A
XPR0008-

Product structure for option retrofit
Item No.

Description

Order code

Option retrofit (please enter the serial number)

XPR0009-_ _Software:Standard
Mathematics

XPR0009-A
XPR0009-B

Option:Standard
Neutral

XPR0009-_ A
XPR0009-_ B

Standard
Modbus TCP slave (Modbus RTU only with RS485)

XPR0009-_ _ A
XPR0009-_ _ C

RS485 interface is required for Modbus RTU. If there is no An
RS485 in the device, a new CPU board must be ordered.

The software option can be enabled directly on the device. After ordering you will
receive instructions and a code that you must enter.

68

Endress+Hauser

Ecograph T, RSG35

Repair

13.3

ReturnThe requirements for safe device return can vary depending on the device type andnational legislation.
1.

Refer to the website for more information:
http://www.endress.com/support/return-material

2.

Return the device if repairs or a factory calibration are required, or if the wrong
device was ordered or delivered.

13.4

Disposal

13.4.1IT security

Observe the following instructions before disposal:
1.

Clear data

2.

Reset the device

3.

Delete/change passwords

4.

Delete user

5.

Carry out alternative or complementary measures to destroy the storage medium

13.4.2

Removing the measuring device

1.

Switch off the device

2.

Carry out the mounting and connection steps from the "Mounting the measuring
device" and "Connecting the measuring device" sections in reverse order. Observe thesafety instructions.

13.4.3

Disposing of the measuring device

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize thedisposal of WEEE as unsorted municipal waste. Do not dispose of products bearing thismarking as unsorted municipal waste. Instead, return them to Endress+Hauser for
disposal under the applicable conditions.

Endress+Hauser

69

Accessories

Ecograph T, RSG35

14

Accessories
Always quote the serial number of the device when ordering accessories! Installation
instructions are provided with the accessory!

Various accessories, which can be ordered with the device or subsequently from Endress+Hauser, are available for the device. Detailed information on the order code in question isavailable from your local Endress+Hauser sales center or on the product page of theEndress+Hauser website: www.endress.com.

14.1

Device-specific accessories

Description

Order No.

"Industrial Grade" SD card, industry standard, 1GB

71213190

Field Data Manager analysis software with SQL database support (1 x workstation
license, Professional version)

MS20-A1

OPC server software (full version on CD)

RXO20-11

Description

Order No.

Accessories for RXU10 data manager

RXU10- _ _Identifier:
Cable set RS232 for connection to PC or modem
USB - RS232 converter
Cable USB-A - USB-B, 1.8 m (5.9 ft)
Configuration software "FieldCare Device Setup" + USB cable

RXU10-B _RXU10-E _RXU10-F _RXU10-G _Description

Order No.

Field housing IP65 (for panel-mounted device)

RXU10-H _254 (10)320 (12.6)

320 (12.6)

A0021773

 1070

Dimensions in mm (in)Endress+Hauser

Ecograph T, RSG35

Accessories

Description

Order No.

Desktop housing (for panel-mounted device), cable with Schuko plug
Desktop housing (for panel-mounted device), cable with US plug
Desktop housing (for panel-mounted device), cable with Swiss plug

RXU10-I _RXU10-J _RXU10-K _236 (9.29)

188 (7.4)

173 (6.81)

293.4 (11.55)

184 (7,24)
207.57 (8.17)

A0021772

 11

Dimensions in mm (in)Version:Standard
NeutralEndress+Hauser

RXU10- _ 1
RXU10- _ 2

71Technical data

Measuring principle

Ecograph T, RSG35

15

Technical data

15.1

Function and system design

Electronic acquisition, display, recording, analysis, remote transmission and archiving of
analog and digital input signals.
The device is intended for installation in a panel or cabinet. There is also the option ofoperating it in a desktop housing or field housing.

Measuring system

Multichannel data recording system with multicolored TFT display (145 mm / 5.7" screensize), galvanically isolated universal inputs (U, I, TC, RTD, pulse, frequency), digital inputs,
transmitter power supply, limit relay, communication interfaces (USB, Ethernet, optional
RS232/485), optionally available with Modbus protocol, 128 MB internal memory,external SD card and USB stick. An Essential Version of the reporting software is included
for SQL-supported data analysis at the PC.
The number of inputs available in the basic device can be individually increased usinga maximum of 3 plug-in cards. The device supplies power directly to connected twowire transmitters. The device is configured and operated via the navigator (jog/shuttledial), via the integrated web server and a PC, or via an external keyboard. Online help
facilitates local operation.

Reliability

Dependability
Depending on the device version, the MTBF is between 52 years and 24 years (calculated
based on SN29500 standard at 40°C)
Serviceability
Battery-backed time and data memory. It is advisable to have the backup battery replaced
by a service technician after 10 years.
Real time clock (RTC)• Configurable summer/normal time changeover• Battery buffer. It is advisable to have the backup battery replaced by a service technician
after 10 years.
• Drift: <10 min./year• Time synchronization possible via SNTP or via digital input.
Standard diagnostic functions as per Namur NE 107
The diagnostic code is made up of the error category as per Namur NE 107 and themessage number.
• Cable open circuit, short-circuit• Incorrect wiring• Internal device errors• Overrange/underrange detection• Ambient temperature out-of-range detection
Device error/alarm relay
One relay can be used as an alarm relay. If the device detects a system error (e.g. hardware
defect) or a malfunction (e.g. cable open circuit), the selected output/relay switches.
This "alarm relay" switches if "F"-type errors occur (F = failure), i.e. "M"-type errors (M=Maintenance required) do not switch the alarm relay.

72

Endress+Hauser

Ecograph T, RSG35

Technical data

SafetyThe tamper-proof recorded data are saved and can be transferred to an external SQL
database for archiving in a way that prevents subsequent manipulation.

15.2
Measured variables

Input

Number of analog universal inputs
Standard version without universal inputs. Optional input cards (slot 1-3) with 4 universal
inputs (4/8/12) each.
Number of digital inputs
6 digital inputs
Number of mathematics channels
4 mathematics channels (optional). Mathematics functions can be freely edited via aformula editor.
Integration of calculated values e.g. for totalization.
Number of limit values
30 limit values (individual channel assignment)
Function of analog universal inputs
You are free to choose between the following measured variables for each universal input:
U, I, RTD, TC, pulse input or frequency input.
Integration of input variable for totalization e.g. flow (m3/h) in quantity (m3).
Calculated process variables
The values of the universal inputs can be used to perform calculations in the mathematics
channels.

Measuring range of analog
universal inputs

According to IEC 60873-1: An additional display error of ±1 digit is permitted for every
measured value.

User-definable measuring ranges per universal input of the multifunction card:
Measuredvariable

Measuring range

Maximum measured error ofmeasuring range (oMR),
temperature drift

Input
resistance

Current (I)

0 to 20 mA; 0 to 20 mA quadratic
0 to 5 mA
4 to 20 mA; 4 to 20 mA quadratic±20 mA
Overrange: up to 22 mA or -22 mA±0.1% oMR
Temperature drift: ±0.01%/K oMR

Load: 50 Ω
±1 Ω

Voltage (U)>1 V

0 to 10 V; 0 to 10 V quadratic
0 to 5 V
1 to 5 V; 1 to 5 V quadratic
±10 V
±30 V

±0.1% oMR
Temperature drift: ±0.01%/K oMR≥1 MΩ

Voltage (U)≤1 V

0 to 1 V; 0 to 1 V quadratic±1 V±150 mV±0.1% oMR
Temperature drift: ±0.01%/K oMR≥2.5 MΩ

Endress+Hauser

73

Technical data

Measuredvariable

Ecograph T, RSG35

Measuring range

Maximum measured error ofmeasuring range (oMR),
temperature drift

Resistance
Pt100: -200 to 850 °C (-328 to 1562 °F) (IEC 60751:2008, α=0.00385)
thermometer Pt100: -200 to 510 °C (-328 to 950 °F) (JIS C 1604:1984, α=0.003916)(RTD)
Pt100: -200 to 850 °C (-328 to 1562 °F) (GOST 6651-94, α=0.00391)
Pt500: -200 to 850 °C (-328 to 1562 °F) (IEC 60751:2008, α=0.00385)
Pt500: -200 to 510 °C (-328 to 950 °F) (JIS C 1604:1984, α=0.003916)
Pt1000: -200 to 600 °C (-328 to 1112 °F) (IEC 60751:2008, α=0.00385)
Pt1000: -200 to 510 °C (-328 to 950 °F) (JIS C 1604:1984, α=0.003916)

4-wire: ±0.1% oMR
3-wire: ±(0.1% oMR + 0.8 K)
2-wire: ±(0.1% oMR + 1.5 K)
Temperature drift: ±0.01%/K oMR

Cu50: -50 to 200 °C (-58 to 392 °F) (GOST 6651-94, α=4260)
Cu50: -200 to 200 °C (-328 to 392 °F) (GOST 6651-94, α=4280)
Pt50: -200 to 1100 °C (-328 to 2012 °F) (GOST 6651-94, α=0.00391)
Cu100: -200 to 200 °C (-328 to 392 °F) (GOST 6651-94, α=4280)

4-wire: ±0.2% oMR
3-wire: ±(0.2% oMR + 0.8 K)
2-wire: ±(0.2% oMR + 1.5 K)
Temperature drift: ±0.02%/K oMR

Pt46: -200 to 1100 °C (-328 to 2012 °F) (GOST 6651-94, α=0.00391)
Cu53: -200 to 200 °C (-328 to 392 °F) (GOST 6651-94, α=4280)

4-wire: ±0.3% oMR
3-wire: ±(0.3% oMR + 0.8 K)
2-wire: ±(0.3% oMR + 1.5 K)
Temperature drift: ±0.02%/K oMR

Thermocoupl Type J (Fe-CuNi): -210 to 1200 °C (-346 to 2192 °F) (IEC 60584:2013)
es (TC)Type K (NiCr-Ni): -270 to 1300 °C (-454 to 2372 °F) (IEC 60584:2013)
Type L (NiCr-CuNi): -200 to 800 °C (-328 to 1472 °F) (GOST R8.585:2001)
Type L (Fe-CuNi): -200 to 900 °C (-328 to 1652 °F) (DIN 43710-1985)
Type N (NiCrSi-NiSi): -270 to 1300 °C (-454 to 2372 °F) (IEC 60584:2013)
Type T (Cu-CuNi): -270 to 400 °C (-454 to 752 °F) (IEC 60584:2013)
Type A (W5Re-W20Re): 0 to 2500 °C (32 to 4532 °F) (ASTME 988-96)
Type B (Pt30Rh-Pt6Rh): 42 to 1820 °C (107.6 to 3308 °F) (IEC 60584:2013)
Type C (W5Re-W26Re): 0 to 2315 °C (32 to 4199 °F) (ASTME 988-96)
Type D (W3Re-W25Re): 0 to 2315 °C (32 to 4199 °F) (ASTME 988-96)
Type R (Pt13Rh-Pt): -50 to 1768 °C (-58 to 3214 °F) (IEC 60584:2013)
Type S (Pt10Rh-Pt): -50 to 1768 °C (-58 to 3214 °F) (IEC 60584:2013)
Pulse input(I) 1)

Min. pulse length 40 μs, max. 12.5 kHz; 0 to 7 mA = LOW; 13 to 20 mA = HIGH

Frequency
input (I) 1)

0 to 10 kHz, overrange: up to 12.5 kHz; 0 to 7 mA = LOW; 13 to 20 mA = HIGH

1)

Input
resistance±0.1% oMR from -100 °C (-148 °F)±0.1% oMR from -130 °C (-202 °F)±0.1% oMR from -100 °C (-148 °F)±0.1% oMR from -100 °C (-148 °F)±0.1% oMR from -100 °C (-148 °F)±0.1% oMR from -200 °C (-328 °F)
Temperature drift: ±0.01%/K oMR≥1 MΩ±0.15% oMR from 500 °C (932 °F)±0.15% oMR from 600 °C (1112 °F)±0.15% oMR from 500 °C (932 °F)±0.15% oMR from 500 °C (932 °F)±0.15% oMR from 100 °C (212 °F)±0.15% oMR from 100 °C (212 °F)
Temperature drift: ±0.01%/K oMR≥1 MΩ

Load: 50 Ω
±1 Ω
±0.02% @ f <100 Hz of reading±0.01% @ f ≥100 Hz of reading
Temperature drift: 0.01% of
measured value over the entire
temperature range

If a universal input is used as a frequency or pulse input, a series resistor must be used in series connection with the voltage source. Example: 1.2kΩ series resistor at 24 V

Maximum load of inputs
Limit values for input voltage and current as well as cable open circuit detection/line influence/temperature compensation:
Measured variable

Limit values (steady-state, without
destroying input)

Cable open circuit detection/line influence/temperature compensation

Current (I)Maximum permitted input voltage: 2.5 V
Maximum permitted input current: 50 mA

4 to 20 mA range with disengageable cable open circuit monitoring to NAMUR
NE43. The following error ranges apply when NE43 is switched on:≤3.8 mA: underrange≥20.5 mA: overrange≤ 3.6 mA or ≥ 21.0 mA: open circuit (display shows: – – – –)Pulse, frequency (I)Maximum permitted input voltage: 2.5 V
Maximum permitted input current: 50 mA

No cable open circuit monitoring

Voltage (U) >1 V

Maximum permitted input voltage: 35 V

1 to 5 V range with disengageable cable open circuit monitoring:<0.8 V or >5.2 V: cable open circuit (display shows: - - - -)Voltage (U) ≤1 V

Maximum permitted input voltage: 24 V

74

Endress+Hauser

Ecograph T, RSG35

Measured variable

Technical data

Limit values (steady-state, without
destroying input)

Cable open circuit detection/line influence/temperature compensation

Resistance
Measuring current: ≤1 mA
thermometer (RTD)

Maximum barrier resistance (or line resistance):
4-wire: max. 200 Ω; 3-wire: max. 40 Ω
Maximum influence of barrier resistance (or line resistance) for Pt100, Pt500
and Pt1000: 4-wire: 2 ppm/Ω, 3-wire: 20 ppm/Ω
Maximum influence of barrier resistance (or line resistance) for Pt46, Pt50,
Cu50, Cu53, Cu100 and Cu500: 4-wire: 6 ppm/Ω, 3-wire: 60 ppm/Ω
Cable open circuit monitoring if any connection is interrupted.

Thermocouples (TC) Maximum permitted input voltage: 24 V

Influence of cable resistance: <0.001%/Ω
Error, internal temperature compensation: ≤ 2 K

Scan rate
Current/voltage/pulse/frequency input: 100 ms per channel
Thermocouples and resistance thermometer: 1 s per channel
Data storage / save cycle
Selectable save cycle. Choose from: 1s / 2s / 3s / 4s / 5s / 10s / 15s / 20s / 30s / 1min /2min / 3min / 4min / 5min / 10min / 15min / 30min / 1h
Typical logging duration
Prerequisites for following tables:• No limit value violation / integration• Digital input not used
• Signal analysis 1: off, 2: day, 3: month, 4: year
• No active mathematics channels
Frequent entries in the event log reduce the memory availability!
128 MB internal memory:
Analog inputs

Channels in groups

Storage cycle (weeks, days, hours)
5 min

1 min

30 s

10 s

1s

1

1/0/0/0

668, 4, 14

135, 0, 5

67, 4, 4

22, 3, 20

2, 1, 18

4

4/0/0/0491, 0, 10

99, 4, 1749, 6, 12

16, 4, 15

1, 4, 168

4/4/0/0

246, 1, 14

49, 6, 1

24, 6, 19

8, 2, 70, 5, 2012

4/4/4/0

164, 2, 4

33, 1, 18

16, 4, 13

5, 3, 210, 3, 21

External memory, 1 GB SD card:
Analog inputs

Channels in groups

Storage cycle (weeks, days, hours)
5 min

1 min

30 s

10 s

1s

1

1/0/0/0

12825, 5, 20

2580, 4, 18

1291, 2, 5

430, 4, 14

43, 0, 12

4

4/0/0/0

8672, 5, 12

1749, 6, 13

875, 6, 13

292, 1, 8

29, 1, 148

4/4/0/04343, 1, 1

875, 1, 17

438, 0, 6

146, 0, 17

14, 4, 7

12

4/4/4/0

2896, 6, 13

583, 3, 21

292, 0, 6

97, 2, 20

9, 5, 4

The available storage capacity of the internal and external memory can be displayedin the main menu under "Diagnostics → Device information → Memory
information". The storage capacity depends on the specific device set-up.

Endress+Hauser

75Technical data

Ecograph T, RSG35

Converter resolution
24 bitTotalization
The interim, daily, monthly and yearly value and the total value can be determined (15digit, 64 bit).
Analysis
Recording of quantity/operating time (standard function), also a min/max/mediananalysis within the set time frame.
Digital inputs

Input level

To IEC 61131-2: logical "0" (corresponds to -3 to +5 V), activation with logical "1"(corresponds to +12 to +30 V)

Input frequency

Max. 25 Hz

Pulse length

Min. 20 ms (pulse counter)

Pulse length

Min. 100 ms (control input, messages, operating time)

Input current

Max. 2 mA

Input voltage

Max. 30 V

Selectable functions• Functions of the digital input: control input, ON/OFF event, pulse counter (15-digit, 64bit), operational time, event+operation time, quantity from time, Modbus slave.
• Functions of the control input: start recording, screensaver on, lock setup, timesynchronization, set point monitoring on/off, block keyboard/navigator, start/stop
analysis.

15.3
Auxiliary voltage output

Galvanic isolation

76

Output

The auxiliary voltage output can be used for loop power supply or to control the digital
inputs. The auxiliary voltage is short-circuit proof and galvanically isolated.
Output voltage

24 VDC ±15%Output current

Max. 250 mA

All inputs and outputs are galvanically isolated from each other and designed for thefollowing testing voltages:
Relay

DigitalinAnalogin

Ethernet RS232/RS485

USB

Auxiliary
voltage output

Relay

500
VDC

2 kVDC

2 kVDC

2 kVDC

2 kVDC

2 kVDC

2 kVDC

Digital in

2 kVDC

Galvanic 500 VDC
ally
connected500 VDC

500 VDC

500 VDC

500 VDC

Analog in

2 kVDC

500 VDC

500 VDC

500 VDC

500 VDC

500 VDC

500 VDC

Ethernet

2 kVDC

500 VDC

500 VDC-

500 VDC

500 VDC

500 VDC

RS232/RS485 2 kVDC

500 VDC

500 VDC

500 VDC-

500 VDC

500 VDC

Endress+Hauser

Ecograph T, RSG35

Technical data

Relay

DigitalinAnalogin

Ethernet RS232/RS485

USB

Auxiliary
voltage output

2 kVDC

500 VDC

500 VDC

500 VDC

500 VDC

Galvanicallyconnected

500 VDC

Auxiliary
2 kVDC
voltage output

500 VDC

500 VDC

500 VDC

500 VDC

500 VDC

-USB

Relay outputs

A mix of low voltage (230 V) and safety extra low voltage (SELV circuits) is not permitted
at the connections of the relay contacts.
Alarm relay
1 alarm relay with changeover contact.
Standard relay
5 relays with NO contact, e.g. for limit value messages (can be configured as NC contact).
Relay switching capacity• Max. switching capacity: 3 A @ 30 V DC
• Max. switching capacity: 3 A @ 250 V AC
• Min. switching load: 300 mW
Switching cycles>105

Cable specification

Cable specification, spring terminals
All connections on the rear of the device are designed as pluggable screw or spring
terminal blocks with reverse polarity protection. This makes the connection very quick andeasy. The spring terminals are unlocked with a slotted screwdriver (size 0).
Please note the following when connecting:• Wire cross-section, auxiliary voltage output, digital I/O and analog I/O: max. 1.5 mm2(14 AWG) (spring terminals)• Wire cross-section, mains: max. 2.5 mm2 (13 AWG) (screw terminals)• Wire cross-section, relays: max. 2.5 mm2 (13 AWG) (spring terminals)• Stripping length: 10 mm (0.39 in)
No ferrules must be used when connecting flexible wires to spring terminals.
Shielding and grounding
Optimum electromagnetic compatibility (EMC) can only be guaranteed if the systemcomponents and, in particular, the lines - both sensor lines and communication lines - areshielded and the shield forms as complete a cover as possible. A shielded line must be used
for sensor lines that are longer than 30 m. A shield coverage of 90% is ideal. In addition,
make sure not to cross sensor lines and communication lines when routing them. Connect
the shield as often as possible to the reference ground to ensure optimum EMC protection
for the different communication protocols and the connected sensors.
To comply with requirements, three different types of shielding are possible:• Shielding at both ends
• Shielding at one end on the supply side with capacitance termination at the device• Shielding at one end on the supply side
Experience shows that the best results with regard to EMC are achieved in most cases ininstallations with one-sided shielding on the supply side (without capacitance termination

Endress+Hauser

77Technical data

Ecograph T, RSG35

at the device). Appropriate internal device wiring measures must be taken to allow
unrestricted operation when EMC interference is present. These measures have beentaken into account for this device. Operation in the event of disturbance variables as per
NAMUR NE21 is thus guaranteed.
Where applicable, national installation regulations and guidelines must be observed
during the installation! Where there are large differences in potential between theindividual grounding points, only one point of the shielding is connected directly with thereference ground.
If the shielding of the cable is grounded at more than one point in systems without
potential matching, mains frequency equalizing currents can occur. These can damage
the signal cable or significantly impact signal transmission. In such cases the shieldingof the signal cable is to be grounded on one side only, i.e. it may not be connected to
the ground terminal of the housing. The shield that is not connected should be
insulated!

15.4

Power supply

1

5

Frequency

Terminal assignment

96A0019304

 12

Supply voltage

Terminals on back of device• Extra-low voltage power supply unit ±24 V AC/DC (-10% / +15%) 50/60Hz• Low voltage power supply unit 100 to 230 V AC (±10%) 50/60Hz
An overload protection element (rated current ≤ 10 A) is required for the power cable.

Power consumption• 100 to 230 V: max. 35 VA• 24 V: max. 24 VA
The power actually consumed depends on the individual operating state and the device
version (LPS, USB, brightness of screen, number of channels, etc). The active power here isapprox. 3 W to 20 W.

Power supply failure

78

Battery-backed time and data memory. The device starts automatically following a power
failure.
Endress+Hauser

Ecograph T, RSG35

Technical data

Electrical connection

Details about the electrical connection: →  13

Device plugs• Panel-mounted device: connected to mains via plug-in screw terminals with reverse
polarity protection• Desktop version (option): connected to mains via IEC connector

Overvoltage protection

To avoid high-energy transients on long signal cables, connect a suitable surge arrester
upstream (e.g. E+H HAW562) in series upstream.

Connection data interface,communication

USB ports (standard):
1 x USB port type A (host)A USB 2.0 port is available on a shielded USB A socket at the front of the device. A USB
stick, for example, can be connected to this interface as a storage medium. An external
keyboard or USB hub may also be connected.
1 x USB port type B (function)
A USB 2.0 port is available on a shielded USB B socket at the front of the device. This canbe used to connect the device for communication with a laptop, for example.
Ethernet interface (standard):
Ethernet interface on back, 10/100 Base-T, plug type RJ45. The Ethernet interface can beused to integrate the device via a hub or switch into a PC network (TCP/ IP Ethernet). A
standard patch cable (e.g. CAT5E) can be used for the connection. Using DHCP, the device
can be fully integrated into an existing network without the need for additional
configuration. The device can be accessed from every PC in the network. Normally only theautomatic assignment of the IP address must be configured at the client. When the device
is started, it can automatically retrieve the IP address, subnet mask and gateway from a
DHCP server. If a DHCP is not used, these settings must be made directly in the device(depends on the network to which the device is to be connected). Two Ethernet function
LEDs are located on the rear of the device.
The following functions are implemented:• Data communication with PC software (analysis software, configuration software, OPC
server)• Web server• WebDAV (Web-based Distributed Authoring and Versioning) is an open standard forthe provisioning of files via the HTTP protocol. The data saved on the device's SD card
can be read out using a PC. A Web browser or a WebDAV client can be selected as the
network drive for this on the PC side.
Serial RS232/RS485 interface (option):
A combined RS232/RS485 connection is available on a shielded SUB D9 socket at the rearof the device. This can be used for data transfer and to connect a modem. For
communication via modem, we recommend an industrial modem with a watchdog
function.
• The following baud rates are supported: 9600, 19200, 38400, 57600, 115200• Max. cable length with a shielded cable: 2 m (6.6 ft) (RS232), or 1000 m (3281 ft)(RS485)
Only one interface can be used at any one time (RS232 or RS485).

Endress+Hauser

79Technical data

Ecograph T, RSG35

15.5
Response time

Input

OutputTime [ms]Current, voltage, pulse

Relay

≤ 550

RTD

Relay

≤ 1150

1)

Relay≤ 1550

Cable open circuit detection, current input

Relay≤ 1150Sensor error RTD, TC

Relay≤ 5000Digital input

Relay

≤ 350

TC

1)

Reference operating
conditions

Performance characteristics

If internal measuring point temperature compensation is used, otherwise values as for voltage

Reference temperature

25 °C (77 °F) ±5 K

Warm-up period

120 min.

Humidity

20 to 60 % rel. humidity

Hysteresis

Can be configured for limit values in the setup

Long-term drift

As per IEC 61298-2: max. ±0.1%/year (of measuring range)

15.6
Mounting location andinstallation dimensions

80

Installation

The device is designed for use in a panel in non-hazardous areas.

Endress+Hauser

Ecograph T, RSG35

Technical data

17 (0.67)

144 (5.67)

144 (5.67)

141 (5.55)

Y34(1.34)

X

A0019301

 13

Panel mounting and dimensions in mm (in)Please observe the installation depth of approx. 158 mm (6.22 in) for the device incl.
terminals and fastening clips.
• Panel cutout: 138 to 139 mm (5.43 to 5.47 in) x 138 to 139 mm (5.43 to 5.47 in)• Panel strength: 2 to 40 mm (0.08 to 1.58 in)• Angle of vision: from the midpoint axis of the display, 75° to the left and right, 65° aboveand below.
• A minimum distance of 15 mm (0.59 in) mm (inch) between the devices must be
observed if aligning the devices in the Y-direction (vertically above one another). A
minimum distance of 10 mm (0.39 in) mm (inch) between the devices must be observed
if aligning the devices in the X-direction (horizontally beside one another).
• Securing to DIN 43 834
Field housing assembly anddesign (optional)

As an option, the device can be ordered ready-mounted in a field housing with IP65.

Desktop housing assemblyand design (optional)

As an option, the device can be ordered ready-mounted in a desktop housing.

Endress+Hauser

Dimensions (B x H x D) approx.: 320 mm (12.6 in) x 320 mm (12.6 in) x 254 mm (10 in)

Dimensions (B x H x D) approx.: 293 mm (11.5 in) x 188 mm (7.4 in) x 211 mm (8.3 in)(dimensions with bracket, feet and installed device)81

Technical data

Ecograph T, RSG35

15.7

Environment

Ambient temperaturerange–10 to +50 °C (14 to 122 °F)

Storage temperature–20 to +60 °C (–4 to +140 °F)

Humidity

5 to 85 %, non-condensing

Climate class

To IEC 60654-1: Class B2

Electrical safety

Class I equipment, overvoltage category II
Pollution level 2

Operating altitude< 2 000 m (6 561 ft) above MSL

Degree of protection

Front of panel-mounted
deviceIP65 / NEMA 4 (UL50 type 4)

Back of panel-mounted
device (terminal side)

IP20

Electromagnetic
compatibility

EMC in accordance with all relevant requirements of the IEC/EN 61326 series and
NAMUR NE21. For details, refer to the Declaration of Conformity.
• Interference immunity: as per IEC/EN 61326 series (industrial environment) / NAMUR
NE21
Maximum measured error <1% of measuring range• Interference emissions: as per IEC 61326-1 Class A

15.8

Mechanical construction

Design, dimensions

Information about design and dimensions →  80

Weight• Panel-mounted device with maximum configuration: approx. 2.2 kg (4.85 lbs)
• Desktop housing (excluding device): approx. 2.3 kg (5 lbs)
• Field housing (excluding device): approx. 4 kg (8.8 lbs)

Materials

Front frame

Zinc die cast GD-Z410, powder-coated

Sight glass

Transparent Makrolon plastic (FR clear 099) UL94-V2

Flap; jog/shuttle dial

Plastic ABS UL94-V2

Mounting guide rail for PCBs; motherboard fixing unit;
display retainer plate

Plastic PA6-GF15 UL94-V2

82

Endress+Hauser

Ecograph T, RSG35

Technical data

Seal to panel wall; seal to display; seal in flap; seal tonavigator

Rubber EPDM 70 Shore A

Casing; rear panel

Galvanized sheet steel St 12 ZE

All materials are silicone-free.
Materials of desktop housing• Housing half-panels: sheet steel, electrolytically plated (powder-coated)• Side sections: aluminum extruded section (powder-coated)• Section ends: colored polyamide

15.9
Operating concept

Display and operating elements

The device can be operated directly onsite, or via remote configuration with the PC via
interfaces and operating tools (Web server, configuration software).
Web server
A Web server is integrated into the device. The Web server offers the following range offunctions:• Easy configuration without additional installed software• Instantaneous value display and diagnostics information• Display of current measured value curves via Web browser (remote control)• Display of historical measured data in numerical format or as a curve• Display of events and logbook entries• Loading/saving of device configuration• Device firmware update• Printout of device configuration
Integrated operating instructions
The device's simple operating concept enables you to perform commissioning for many
applications without the need for hard copy operating instructions. The device has anintegrated help function and displays operating instructions directly on screen if thenavigator (jog/shuttle dial) is pressed for longer than 3 seconds.

Local operation

Display elements
Type
TFT color display
Size (diagonal screen measurement)
145 mm (5.7")
Resolution
VGA 307,200 pixels (640 x 480 pixels)
Backlight
70,000 h half value time (= half brightness)
Number of colors
262,000 viewable colors, 256 colors usedViewing angle
Viewing angle: 130° vertical, 150° horizontal

Endress+Hauser

83Technical data

Ecograph T, RSG35

Screen display• Background color white• Active channels can be assigned to up to 4 groups. These groups can be given a namee.g. "Temp. boiler 1" or "Daily averages" so that they can be uniquely identified.
• Linear scaling• Measured value history: quick retrieval of historical data with zoom function• Preformatted screen displays such as horizontal or vertical curve presentation, bar graphdisplay or digital display.
Operating elements
11

10

9

12
8

7

6 5 4

3

2

1
A0020602-EN

 14

84

Front of device with open flap

Item
No.

Operating function (display mode = display of measured values)(Setup mode = operation in the Setup menu)

1"Navigator": jog/shuttle dial for operating with additional press/hold function.
In display mode: turn the dial to switch between the various signal groups. Press the dial to display themain menu.
In setup mode or in a selection menu: turn the dial counterclockwise to move the bar or the cursor
upwards or to the left, changes the parameter. Turning clockwise moves the bar or cursor down orclockwise, changes the parameter.

2

LED at SD slot. Orange LED lights up or flashes when the device writes to or reads from the SD card.
Do not remove the SD card if the LED is lit or flashing! Risk of data loss!

3

Slot for SD card

4

USB B socket "Function" e.g. to connect to PC or laptop

5Green LED lit: power supply present

6

USB A socket "Host" e.g. for USB memory stick or external keyboard

7

In display mode: alternating status display (e.g. set zoom range) of the analog or digital inputs in theappropriate color of the channel.
In setup mode: different information can be displayed here depending on the display type.

8

In display mode: window for measured value display (e.g. curve display).
In setup mode: display of operating menu

Endress+Hauser

Ecograph T, RSG35

Technical data

Item
No.

Operating function (display mode = display of measured values)(Setup mode = operation in the Setup menu)

9

In display mode: current group name, type of evaluation
In setup mode: name of the current operating item (dialog title)

10

In display mode: displays current date/time
In setup mode: --11

In display mode: alternating display indicating the percentage space on the SD card or USB stick thathas already been used.
Status symbols are also displayed in alternation with the memory information.
In setup mode: the current "direct access" operating code is displayed

12

In display mode: display of current measured values and the status in the event of an error/alarm
condition. In the case of counters, the type of counter is displayed as a symbol.
point has limit value status, the corresponding channel identifier is highlighted
 inIf aredmeasuring(quick detection of limit value violations). During a limit value violation and device
operation, the acquisition of measured values continues uninterrupted.

Languages

The following languages can be selected in the operating menu: German, English, Spanish,French, Italian, Dutch, Swedish, Polish, Portuguese, Czech, Russian, Japanese, Chinese(Traditional), Chinese (Simplified)

Remote operation

Device access via operating tools
Device configuration and measured value retrieval can also be done via interfaces. The
following operating tools are available for this purpose:

System integration

Operating tool

Functions

Access via"Field Data Manager(FDM)" analysis
software, SQL
database support(included in thedelivery)• Export of saved data (measured values, analyses, event log) RS232/RS485, USB,
• Visualization and processing of saved data (measured
Ethernetvalues, analyses, event log)
• Safe archiving of exported data in a SQL database

Web server(integrated into thedevice; access via
browser)• Display of current and historical data and measured value
curves via the Web browser• Easy configuration without additional installed software• Remote access to device and diagnostic information

Ethernet, or Ethernet via
USB

OPC server(optional)

The following instantaneous values can be provided:• Analog channels• Digital channels• Mathematics• Totalizer

RS232/RS485, USB,
Ethernet"FieldCare / DeviceCare" configuration
software• Device configuration• Loading and saving device data (upload/download)• Documentation of the measuring point

USB, Ethernet

The device has (optional) fieldbus interfaces for exporting process values. Measured values
and statuses can also be transmitted to the device via fieldbus. Alarms or errors in thecontext of data transmission are displayed depending on the bus system (e.g. status byte).
The process values are transferred in the same units that are used for display at the device.
Ethernet
The following functions are implemented:• Data communication with PC software (analysis software, configuration software, OPC
server)• Web serverEndress+Hauser

85

Technical data

Ecograph T, RSG35

Modbus RTU/TCP slave
The device can be connected to a Modbus system via RS485 or Ethernet interface. Up to 12
analog inputs and 6 digital inputs can be transmitted via Modbus and stored in the device.

15.10 Certificates and approvals
For certificates and approvals valid for the device: see the data on the nameplate
Approval-related data and documents: www.endress.com/deviceviewer → (enter theserial number)
CE mark

The product meets the requirements of the harmonized European standards. As such, itcomplies with the legal specifications of the EC directives. The manufacturer confirms
successful testing of the product by affixing to it the CE-mark.

Other standards andguidelines• IEC 60529:
Degrees of protection provided by enclosures (IP code)• IEC/EN 61010-1:
Safety requirements for electrical equipment for measurement, control, and laboratoryuse• IEC/EN 61326 series:
Electromagnetic compatibility (EMC requirements)

15.11 Ordering information
Ordering information

Detailed ordering information is available for your nearest sales organizationwww.addresses.endress.com or in the Product Configurator under www.endress.com :1.

Click Corporate

2.

Select the country

3.

Click Products

4.

Select the product using the filters and search field

5.

Open the product page

The Configuration button to the right of the product image opens the ProductConfigurator.
Product Configurator - the tool for individual product configuration• Up-to-the-minute configuration data
• Depending on the device: Direct input of measuring point-specific information suchas measuring range or operating language• Automatic verification of exclusion criteria• Automatic creation of the order code and its breakdown in PDF or Excel output
format• Ability to order directly in the Endress+Hauser Online Shop
Scope of delivery

86

The scope of delivery of the device comprises:• Device (with terminals, as per order)• Panel-mounted device: 2 screw fastening clips• USB cable• Optional: industrial grade SD card (card is located in the SD slot behind the flap on thefront of the device)

Endress+Hauser

Ecograph T, RSG35

Technical data

• "Field Data Manager (FDM)" analysis software on DVD (Essential, Demo or Professionalversion, depending on order)• Delivery note
• Multilanguage Brief Operating Instructions, hard copy

15.12 Supplementary documentation
Standard documentation• Technical Information for Ecograph T RSG35: TI01079R• Operating Instructions for Ecograph T RSG35: BA01146R• Brief Operating Instructions for Ecograph T RSG35: KA01132R• System components and data manager - solutions to complete your measuring point:
FA00016K
Supplementary device-dependent documentation
Operating Instructions for Ecograph T RSG35 with Modbus RTU / TCP slave: BA01258R

Endress+Hauser

87Appendix

Ecograph T, RSG35

16

Appendix

16.1

Operating items in the "Expert" menu

The parameter groups for the Expert setup contain all the parameters of the operating
menus: System, Input and Output Setup, Communication, Application, Diagnostics as wellas other parameters that are reserved for experts only.
For most settings, the "Setup" or "Expert" menu must be quit before the settings are
adopted. However settings such as the date/time are accepted immediately.

Direct access

Navigation

Expert → Direct access

Description

Direct access to active operating items (rapid access). Entering the direct access code takes
you directly to the desired operating parameter. The direct access code is displayed in the
Setup menu on the top right of the display (e.g. 00000-000).

Text entry(e.g. 00000-000)

16.1.1"System" submenu

Basic settings that are needed to operate the device (e.g. date, time, etc.)

Language

Navigation

Expert → System → Language
Direct access code: 010000-000

Description

Select the operating language of the device.

Selection

German, English, Spanish, French, Italian, Dutch, Polish, Portuguese, Russian, Swedish,Czech, Japanese, Chinese (Simplified), Chinese (Traditional)

Factory setting

English; or preset to customer's preferred language

Device tag

Navigation

Expert → System → Device tag
Direct access code: 000031-000

Description

Individual device tag

User entry

Text entry (max. 32 characters)

Factory setting

Unit 1

88

Endress+Hauser

Ecograph T, RSG35

Appendix

Temp. unit

Navigation

Expert → System → Temp. unit
Direct access code: 100001-000

Description

Selection of temperature unit. All directly connected thermocouples or resistancethermometers (RTD) are displayed in the preset engineering units.

Selection°C, °F, K

Factory setting°C

Decimal separator

Navigation

Expert → System → Decimal separator
Direct access code: 100003-000

Description

Select in which form the decimal separator character is to be displayed.

Selection

Comma, point

Factory setting

Comma

Fault switching

Navigation

Expert → System → Fault switching
Direct access code: 100002-000

Description

If the device detects a system error (e.g. hardware defect) or a fault (e.g. cable open
circuit), the selected output switches.

Selection

Not used, Relay x
All the available relays are displayed.

Factory setting

Relay 1

Keyboard layout

Navigation

Description

Endress+Hauser

Expert → System → Keyboard layout
Direct access code: 100020/000
Please select the keyboard layout. Only relevant if external keyboard is used.

89Appendix

Ecograph T, RSG35

Selection

Germany, Switzerland, France, USA, USA International, UK, Italy

Factory setting

Germany

Swap mouse buttons

Navigation

Expert → System → Swap mouse buttons
Direct access code: 100050/000

Description

Swap function of left and right mouse button.

Selection

No, Yes

Factory setting

No

Paper size

Navigation

Expert → System → Paper size
Direct access code: 540004/000

Description

Please select the paper size of the printer connected to the PC.

Selection

DIN A4, US Letter

Factory setting

DIN A4Lock operation

Navigation

Expert → System → Lock operation
Direct access code: 100060/000

Description

Local operation is locked in cases of inactivity once the set time has elapsed to prevent
inadvertent operation (e.g. when cleaning the device). The device is unlocked by pressing
the navigator or the OK operating key for 3 s. When using an external keyboard, thedevice is unlocked with the key combination "Ctrl-Alt-Del".

Selection

Never, After 2 (5, 10, 15) minutes

Factory setting

After 5 minutes

PRESET

90

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Description

Expert → System → PRESET
Direct access code: 000044-000
Caution: Resets all the parameters to the factory settings!
Only visible/editable if the service code has been entered.

Selection

No, Factory reset, Customer setting

Clear memory

Navigation

Expert → System → Clear memory
Direct access code: 059000-000

Description

Delete internal memory

Selection

No, Yes

Confirm deletion

Navigation

Expert → System → Confirm deletion
Direct access code: 059001-000

Description

Please confirm that you want to clear the memory.

Selection

No, Yes

Factory setting

No"Date/time setup" (submenu)
Navigation
Description

Expert → System → Date/time setup
Contains settings for date/time.

Date format

Navigation

Expert → System → Date/time setup → Date format
Direct access code: 110000-000

Description

Select in which format the date is to be set and displayed.

Selection

DD.MM.YYYY, MM/DD/YYYY, YYYY-MM-DD

Endress+Hauser

91

Appendix

Factory setting

Ecograph T, RSG35

DD.MM.YYYYTime format

Navigation

Expert → System → Date/time setup → Time format
Direct access code: 110001-000

Description

Select in which format the time is to be set and displayed.

Selection

24 hour, 12 hour AM/PM

Factory setting

24 hour"Date/time" submenu
Navigation
Description

Expert → System → Date/time setup → Date/time
Contains parameters for setting the date/time.

UTC time zone

Navigation

Expert → System → Date/time setup → Date/time → UTC time zone
Direct access code: 120000-000

Description

Display of the current UTC time zone is on (UTC = universal time coordinated).

Current date/time

Navigation

Description

Expert → System → Date/time setup → Date/time → Current date/time
Direct access code: 120003-000
Displays the current date and the current time.

"Change date/time" submenu
Description
Navigation

Contains parameters for changing the date/time.
Expert → System → Date/time setup → Change date/time

UTC time zone
92

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → System → Date/time setup → Date/time → Change date/time → UTC timezone
Direct access code: 120010-000

Description

Set your UTC time zone (UTC = universal time coordinated).

Selection-12:00, -11:00: Samoa, -10:00: Hawaii, -09:30: Marquesas, -09:00: Alaska, -08:00: LA,-07:00: Denver, -06:00: Chicago, -05:00: New York, -04:00: Caracas, -03:30: St.John’s,-03:00: Brasilia, -02:00: Atlantic, -01:00: Azores, +00:00: London, +01:00: Berlin,+02:00: Cairo, +03:00: Moscow, +03:30: Tehran, +04:00: Abu Dhabi, +04:30: Kabul,+05:00: Islamabad, +05:30: New Delhi, +05:45: Kathmandu, +06:00: Dhaka, +06:30:
Pyinmana, +07:00: Bangkok, +08:00: Peking, +08:45, +09:00: Tokyo, +09:30: Adelaide,+10:00: Canberra, +10:30: Lord-Howe, +11:00:Solom.Isl., +11:30: Norfolk, +12:00:
Auckland, +12:45: Chatham, +13:00, +14:00

Date/time

Navigation

Expert → System → Date/time setup → Date/time → Change date/time → Date/time
Direct access code: 120013-000

Description

Set the current date and time for the unit here.

User entry

Date/time in set format"NT/ST changeover" submenu
Navigation
Description

Expert → System → Date/time setup → NT/ST changeover
Contains settings for normal time/summer time changeover.

NT/ST changeover

Navigation

Expert → System → Date/time setup → NT/ST changeover → NT/ST changeover
Direct access code: 110002-000

Description

Function for summer/normal time changeover.
Automatic: Changes to the local regional regulations;
Manual: Changeover times can be set in the following addresses;
Off: No changeover times required.

Selection

Off, Manual, Automatic

Factory setting

Automatic

NT/ST region

Endress+Hauser

93

Appendix

Ecograph T, RSG35

Navigation

Expert → System → Date/time setup → NT/ST changeover → NT/ST region
Direct access code: 110003-000

Description

Selects the regional settings for summer/normal time changeover.
Only visible if NT/ST changeover = Automatic.

Selection

Europe, USA

Factory setting

EuropeBegin summer time

Occurrence

Navigation

Expert → System → Date/time setup → NT/ST changeover → Occurrence
Direct access code: 110005-000

Description

Day, when in the spring a change from normal to summer time occurs.
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

Selection1., 2., 3., 4., Last

Factory setting

Last

Day

Navigation

Expert → System → Date/time setup → NT/ST changeover → Day
Direct access code: 110006-000

Description

Day, when in the spring a change from normal to summer time occurs.
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

Selection

Sunday, Monday, Tuesday, Wednesday, Thursday, Friday, Saturday

Factory setting

SundayMonthNavigation

94

Expert → System → Date/time setup → NT/ST changeover → Month
Direct access code: 110007-000

Endress+Hauser

Ecograph T, RSG35

Appendix

Description

Month, when in the spring a change from normal to summer time occurs.
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

Selection

January, February, March, April, May, June, July, August, September, October, November,
December

Factory setting

March

Date

Navigation

Expert → System → Date/time setup → NT/ST changeover → Date
Direct access code: 110008-000

Description

Date next spring when a change from normal to summer time occurs.
Only visible if NT/ST changeover = Automatic or Manual. Cannot be edited.

Time

Navigation

Expert → System → Date/time setup → NT/ST changeover → Time
Direct access code: 110009-000

Description

Time at which the changeover from normal time to summer time is forwarded by one hour
(format: hh:mm).
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

User entry

Time in set time format

Factory setting

02:00

End summer time

Occurrence

Navigation

Expert → System → Date/time setup → NT/ST changeover → Occurrence
Direct access code: 110011-000

Description

Day, when in the autumn a change from summer to normal time occurs.
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

Selection1., 2., 3., 4., Last

Endress+Hauser

95

Appendix

Factory setting

Ecograph T, RSG35

Last

Day

Navigation

Expert → System → Date/time setup → NT/ST changeover → Day
Direct access code: 110012-000

Description

Day, when in the autumn a change from summer to normal time occurs.
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

Selection

Sunday, Monday, Tuesday, Wednesday, Thursday, Friday, Saturday

Factory setting

SundayMonthNavigation

Expert → System → Date/time setup → NT/ST changeover → Month
Direct access code: 110013-000

Description

Month, when in the autumn a change from summer to normal time occurs.
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

Selection

January, February, March, April, May, June, July, August, September, October, November,
December

Factory setting

October

Date

Navigation

Expert → System → Date/time setup → NT/ST changeover → Date
Direct access code: 110014-000

Description

Date next autumn when summer time changes back to normal time.
Only visible if NT/ST changeover = Automatic or Manual. Cannot be edited.

Time

Navigation

96

Expert → System → Date/time setup → NT/ST changeover → Time
Direct access code: 110015-000

Endress+Hauser

Ecograph T, RSG35

Appendix

Description

Time at which the changeover from summer time to normal time is turned back by one
hour (in the set time format).
Visible for NT/ST changeover = Automatic or Manual. Only editable if NT/ST changeover =Manual.

User entry

Time in set time format

Factory setting

02:00"SNTP" submenu
Navigation
Description

Expert → System → Date/time setup → SNTP
Contains settings for time synchronization using the Simple Network Time Protocol(SNTP).

SNTP

Navigation

Expert → System → Date/time setup → SNTP
Direct access code: 110020-000

Description

If switched on, time synchronization is carried out via SNTP once a day.
Note: Only possible via Ethernet.
Port 123 must be open in the firewall. The user/network administrator is responsible forthe accuracy of the time server.

Selection

No, Yes

Factory setting

No

SNTP server 1

Navigation

Expert → System → Date/time setup → SNTP server → SNTP server 1
Direct access code: 110021-000

Description

Please specify the address of the time server (or the IP address).
Note: The DNS server must be configured (see Communication/Ethernet).
Your administrator can provide the address where necessary.

User entry

Text field

SNTP server 2

Navigation

Endress+Hauser

Expert → System → Date/time setup → SNTP server → SNTP server 2
Direct access code: 110025-000
97Appendix

Description

Ecograph T, RSG35

Shows the IP address of the time server if it was automatically allocated via DHCP. Noneditable display text.
An attempt is always made to synchronize the time via SNTP server 1 first (provided
it is configured)
DHCP must be switched on (see Communication/Ethernet).
DHCP server: Option 42"Security" submenu
Navigation
Description

Expert → System → Security
Contains settings that protect the device against unauthorized operation andconfiguration.

Protected by

Navigation

Expert → System → Security → Protected by
Direct access code: 100006-000

Description

Configure how the device should be protected.

Selection

Open access, access code, user roles

Factory setting

Open access

Access code

Navigation

Expert → System → Security → Access code
Direct access code: 100000-000

Description

Using this code, setup access can be protected from unauthorized persons. In order tochange any parameter the correct code must be entered. Factory setting: "0", i.e. changes
can be made at any time.
Hint: Make a note of the code and store in a safe place.
Only visible if "Protected by" = "Access code"

User entry

4-digit number

Factory setting0

Set point code

Navigation

98

Expert → System → Security → Set point code
Direct access code: 100030-000
Endress+Hauser

Ecograph T, RSG35

Description

Appendix

If the unit is protected by an access code a set point code can also be defined. The user canchange the set points once the set point code is entered. All other operating positions
remain locked, however.
Only visible if an activation code has been defined.
Factory default: "0" means that alarm set points can only be changed by entering the accesscode.
Alarm set point code and access code should not be identical!

User entry

4-digit number

Factory setting0

Lock hardware

Navigation

Description

Expert → System → Security → Lock hardware
Direct access code: 100099-000
Device functions/interfaces that are not used can be switched off for security reasons.
Fieldbus systems may also be affected in the case of Ethernet or the serial interface.
Please follow the operating instructions.

Selection

Panel version: Ethernet (all ports/services), USB A socket front, USB A socket back, USB B
socket front, serial interface, SD card

Factory setting

No lock"Authentication" submenu
Navigation
Description

Expert → System → Security → Authentication
Define the passwords for the various user roles to allow device access.
Only visible if "Protected by" = "User roles"

Operator
ID: operator
Password

Navigation

Expert → System → Security → Authentication → Password
Direct access code: 470105/000

Description

Enter a password for this user account.

Selection

Text entry max. 12 characters

Factory setting

operator

Endress+Hauser

99

Appendix

Ecograph T, RSG35

Administrator
ID: admin
Password

Navigation

Expert → System → Security → Authentication → Password
Direct access code: 470102/000

Description

Enter a password for this user account.

Selection

Text entry max. 12 characters

Factory setting

adminService
ID: service
Password

Navigation

Expert → System → Security → Authentication → Password
Direct access code: 470101/000

Description

Enter a password for this user account.

Selection

Text entry max. 12 characters

Factory setting

service"External memory" submenu
Navigation
Description

Expert → System → External memory
Settings for the external data carrier, amongst other things which data is to be stored inwhich format on the external data carrier.

Save asNavigation

Expert → System → External memory → Save as
Direct access code: 140000-000

Description"Protected format": All data is stored in a manipulation protected encrypted format. Thisdata can only be visualized by the PC analysis software supplied.
"Open format": Data is stored in a CSV format, this can be opened by a number of different
programs (for e.g. MS Excel) (Attention: no manipulation security).

Selection

Protected format, Open format (*.csv)

Factory setting

Protected format

100

Endress+Hauser

Ecograph T, RSG35

Appendix

SD cardMemory build-up

Navigation

Description

Expert → System → External memory → Memory build-up
Direct access code: 140001-000"Stack memory": No more data can be stored once the data carrier is full.
"Ring memory": Once the data carrier is full the oldest data is deleted so that new data canbe stored (First in first out (FIFO).
The "Ring memory" setting refers only to the automatic storage of measured values.
Manual save functions ("Operation -> SD card -> Update/Save measured values") are
not affected.

Selection

Stack memory, Ring memory (FIFO)"Ring memory" can be selected only if "Save as" is set to "Protected format" (and not"CSV").

Factory setting

Stack memory

Warning at

Navigation

Description

Expert → System → External memory → Warning at
Direct access code: 140005-000
Issues a warning before the data carrier is x% full.
A warning is indicated on the device and this is also stored in the event buffer.
A relay can also be switched.
Only for external SD card (does not apply to USB stick)!

User entry

0 to 99%

Factory setting90

Switches relay

Navigation

Description

Endress+Hauser

Expert → System → External memory → Switches relay
Direct access code: 140006-000
When the warning "Data carrier full" is displayed a relay can also be switched on.

101

Appendix

Ecograph T, RSG35

Selection

Not used, Relay x
All the available relays are displayed.

Factory setting

Not used

CSV settings
Also configurable if "Protected format" is set.

Separator for CSV

Navigation

Expert → System → External memory → Separator for CSV
Direct access code: 140002-000

Description

Configure which separator is used by your application (e.g. in Excel = semicolon).

Selection

Comma, Semicolon

Factory setting

Semicolon

Date/time

Navigation

Expert → System → External memory → Date/time
Direct access code: 140003-000

Description

Please select if the date and time should be stored in one column or separate columns
when data is saved in CSV format files.

Selection

One column, Separate columns

Factory setting

Separate columns

Operational time

Navigation

Expert → System → External memory → Operational time
Direct access code: 140004-000

Description

Please select in which format operation times will be stored/displayed.

Selection

0 seconds, 0.0000 hours, 0.00000 days, 0000h00:00

Factory setting

0000h00:00

102

Endress+Hauser

Ecograph T, RSG35

Appendix"Messages" submenu
Navigation
Description

Expert → System → Messages
Contains settings for displaying/acknowledging messages. Examples of messages include:
messages triggered by set points; messages triggered by a digital input; error messages;etc.

Acknowledging messages

Navigation

Expert → System → Messages → Acknowledging messages
Direct access code: 100040-000

Description

The time the message is acknowledged can be saved in the events list.

Selection

Do not save, Save

Factory setting

Do not save

Switches relay

Navigation

Expert → System → Messages → Switches relay
Direct access code: 100042-000

Description

A relay can be switched as soon as a message that has to be confirmed is displayed (e.g.
on/off message, device errors etc.).
The relay takes the initial state as soon as all messages have been confirmed.

Selection

Not used, Relay x
All the available relays are displayed.

Factory setting

Not used"Screensaver" submenu
Navigation
Description

Expert → System → Screensaver
To increase the life span of the LCD the rear illumination can be switched off (=screensaver).

Screensaver

Navigation

Endress+Hauser

Expert → System → Screensaver → Screensaver
Direct access code: 160000-000
103Appendix

Ecograph T, RSG35

Description"Switched off": LCD is always switched on.
“Switch on for x min.”: Display goes dark after x minutes. All other functions remain in
operation. Press an operating key: Illumination is switched back on.
"Switched daily": Enter time span.

Selection

Switched off, On after 10 min, On after 30 min, On after 60 min, Switched daily, Control
input

Factory setting

Switched off
This setting has no effect if the screensaver is controlled by a digital input.

ON daily from

Navigation

Description

Expert → System → Screensaver → ON daily from
Direct access code: 160001-000
Set time (hh:mm) as of when the screensaver should be switched on (e.g. shift end time).
The screensaver is switched off as soon as the device is operated via onsite operation.
After 1 min of inactivity it switches back on automatically.
Only visible if screensaver = Switched daily

User entry

Time (hh:mm)

Factory setting

20:00

OFF daily from

Navigation

Expert → System → Screensaver → OFF daily from
Direct access code: 160002-000

Description

Set time (hh:mm) as of when the screensaver should be switched off (e.g. shift start time).
Only visible if screensaver = Switched daily

User entry

Time (hh:mm)

Factory setting

07:00

Alarm response

Navigation

104

Expert → System → Screensaver → Alarm response
Direct access code: 160003-000

Endress+Hauser

Ecograph T, RSG35

Description

Appendix"Off on alarm": If limit value violations occur or status signal "Maintenance required (Mxxx)"
or "Function check (Cxxx)" is active, the screensaver will be automatically deactivated.
"Always on": If limit value violations occur or status signal "Maintenance required (Mxxx)"
or "Function check (Cxxx)" is active, the screensaver will not be deactivated.
Active messages that require acknowledgment or an active status signal "Failure(Fxxx)" or "Out of specification (Sxxx)" deactivate the screensaver always.

Selection

Off on alarm, Always on

Factory setting

Off on alarm"Device options" submenu
Navigation

Expert → System → Device options

Description

Hardware and software options of the device.

Activation code

Navigation

Description

Expert → System → Device options → Activation code
Direct access code: 000057-000
Here, you can enter a code to enable the device options.
The options that can be retrofitted are listed under "spare parts" →  68
Note: When an activation code is entered, the device is restarted in order to enable thenew option.
• The activation code entered is not displayed, i.e. this parameter is always empty
following a restart.
• Pay attention to case-sensitivity.

User entry

Text

Slot 1

Navigation

Description

Expert → System → Device options → Slot 1
Direct access code: 990000-000
Shows hardware or software options.
Cannot be edited.
The assignment can be specified in the PC operating software for offline
configuration.

Selection

Endress+Hauser

Not assigned, Universal inputs

105

Appendix

Ecograph T, RSG35

Slot 2

Navigation

Description

Expert → System → Device options → Slot 2
Direct access code: 990001-000
Shows hardware or software options.
Cannot be edited.
The assignment can be specified in the PC operating software for offline
configuration.

Selection

Not assigned, Universal inputs

Slot 3

Navigation

Description

Expert → System → Device options → Slot 3
Direct access code: 990002-000
Shows hardware or software options.
Cannot be edited.
The assignment can be specified in the PC operating software for offline
configuration.

Selection

Not assigned, Universal inputs

Communication

Navigation

Expert → System → Device options → Communication
Direct access code: 990006-000

Description

Shows hardware or software options.
Cannot be edited.

Selection

USB + Ethernet, USB + Ethernet + RS232/485

Fieldbus

Navigation

Expert → System → Device options → Fieldbus
Direct access code: 990005-000

Description

Shows hardware or software options.
Cannot be edited.

Selection

Not available, Modbus Slave

106

Endress+Hauser

Ecograph T, RSG35

Appendix

Application

Navigation

Expert → System → Device options → Application
Direct access code: 990007-000

Description

Shows hardware or software options.
Cannot be edited.

Selection

Standard, Maths

16.1.2"Inputs" submenu

Settings for the analog and digital inputs.

"Universal inputs" submenu
Navigation
Description

Expert → System → Inputs → Universal inputs
Settings for the connected measuring points.

Add input

Navigation

Expert → System → Inputs → Universal inputs → Add input
Direct access code: 222000/000

Description

Addition of an input that must be switched on and configured according to the input
signal.

Selection

No, Universal input x

Factory setting

No

Delete input

Navigation

Expert → System → Inputs → Universal inputs → Delete input
Direct access code: 222001/000

Description

Delete an input configuration.

Selection

No, Universal input x

Factory setting

No

Endress+Hauser

107Appendix

Ecograph T, RSG35"Universal input x" submenu
Navigation
Description

Expert → System → Inputs → Universal inputs → Universal input x
View or change settings for the selected channel.
x = place holder for selected universal input

SignalNavigation

Expert → Inputs → Universal inputs → Universal input x → Signal
Direct access code: 220000-0xx
Examples: Universal input 1: 220000-000; Universal input 12: 220000-011

Description

Select the type of signal connected (current, voltage, etc.). The channel is switched off if nosignal type is selected (factory default).

Selection

Switched off, Current, Voltage, Resistance thermometer, Thermocouple, Pulse counter,
Frequency input, Modbus Slave (option)

Factory setting

Switched off

Range

Navigation

Expert → Inputs → Universal inputs → Universal input x → Range
Direct access code: 220001-0xx
Examples: Universal input 1: 220001-000; Universal input 12: 220001-011

Description

Select the input range or which resistance thermometer/thermocouple is connected. The
respective terminal layout can be found in the operating manual or on the rear of the unit.
Only visible if signal ≠ Switched off

Selection

Switched off
Current: 4-20 mA, 0-20 mA, 0-5 mA, 0-20 mA squared, 4-20 mA squared, ±20 mA
Voltage: 0-1 V, 0-10 V, 0-5 V, 1-5 V, ±150 mV, ±1 V, ±10 V, ±30 V, 0-1 V squared, 0-10
V squared, 1-5 V squared
Resistance thermometer: Pt100 (IEC), Pt100 (JIS), Pt100 (GOST), Pt500 (IEC), Pt500 (JIS),
Pt1000 (IEC), Pt1000 (JIS), Pt46 (GOST), Pt50 (GOST), Cu50 (GOST, a=4260), Cu50
(GOST, a=4280), Cu53 (GOST, a=4280), Cu100 (GOST, a=4280)
Thermocouple: Type A (W5Re-W20Re), type B (Pt30Rh-Pt6Rh), type C (W5Re-W26Re),
type D (W3Re-W25Re), type J (Fe-CuNi), type K (NiCr-Ni), type L (Fe-CuNi), type L (NiCrCuNi, GOST), type N (NiCrSi-NiSi), type R (Pt13Rh-Pt), type S (Pt10Rh-Pt), type T (CuCuNi)
Pulse counter
Frequency input
Modbus (option)

Factory setting

Switched off

108

Endress+Hauser

Ecograph T, RSG35

Appendix

Connection

Navigation

Expert → Inputs → Universal inputs → Universal input x → Connection
Direct access code: 220002-0xx
Examples: Universal input 1: 220002-000; Universal input 12: 220002-011

Description

Specify whether RTDs are connected as 2-, 3- or 4-wire systems.
Only visible if signal = Resistance thermometer

Selection

2-wire, 3-wire, 4-wireFactory setting

4-wireChannel ident.

Navigation

Expert → Inputs → Universal inputs → Universal input x → Channel ident.
Direct access code: 220003-0xx
Examples: Universal input 1: 220003-000; Universal input 12: 220003-011

Description

Name of the measuring point connected to this input.
Only visible if signal ≠ Switched off

User entry

Text (16 characters)

Factory setting

Channel x

Plot type

Navigation

Expert → Inputs → Universal inputs → Universal input x → Plot type
Direct access code: 220016-0xx
Examples: Universal input 1: 220016-000; Universal input 12: 220016-011

Description

The analog inputs are scanned in a 100ms cycle. Dependent on the store cycle the selected
data is selected, stored and displayed from the scanned values.

Selection

Instantaneous value, Average, Minimum value, Maximum value, Minimum + Maximum,
Counter, Current value + Counter

Factory setting

AverageTime base

Endress+Hauser

109

Appendix

Ecograph T, RSG35

Navigation

Expert → Inputs → Universal inputs → Universal input x → Time baseDirect access code: 220025-0xx
Examples: Universal input 1: 220025-000; Universal input 12: 220025-011

Description

A current value can be determined from the counter reading with the aid of the time basee.g. input liters, time base = second → current value = liters/second.
Only visible if signal = "Pulse counter" and plot type = "Current value + Counter"

Selection

Second (s), Minute (min), Hour (h), Day (d)Factory setting

Second (s)Engineering unit

Navigation

Expert → Inputs → Universal inputs → Universal input x → Engineering unit
Direct access code: 220004-0xx
Examples: Universal input 1: 220004-000; Universal input 12: 220004-011

Description

Specify the technical (physical) unit for the measuring point connected to this input.
Only visible if signal ≠ Switched off

User entry

Text (6 characters)

Unit/dimension counter

Navigation

Expert → Inputs → Universal inputs → Universal input x → Unit/dimension counter
Direct access code: 220024-00x
Examples: Universal input 1: 220024-000; Universal input 12: 220024-011

Description

Technical unit of the count input, e.g. gal, cf, ... .
Only visible if signal = "Pulse counter" and plot type = "Current value + Counter"

User entry

Text (max. 6 characters)

Pulse counter

Navigation

Expert → Inputs → Universal inputs → Universal input x → Pulse counter
Direct access code: 220017-0xx
Examples: Universal input 1: 220017-000; Universal input 12: 220017-011

Description

Specify whether the pulse counter is a fast or slow (up to max. 25 Hz) counter. For
example if the number of state changes from a relay are to be monitored then you mustset "up to 25Hz".
Only visible if signal = Pulse counter

Selection

Up to 13kHz, Up to 25Hz

110

Endress+Hauser

Ecograph T, RSG35

Factory setting

Appendix

Up to 13kHz

Pulse value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Pulse value
Direct access code: 220010-0xx
Examples: Universal input 1: 220010-000; Universal input 12: 220010-011

Description

Factor, that when multiplied by the input signal results in the required physical value.
Example: 1 pulse equals 5 m3 -> enter a "5".
Only visible if signal = Pulse counter

User entry

Number, max. 8 digits

Factory setting

1

Decimal point

Navigation

Expert → Inputs → Universal inputs → Universal input x → Decimal point
Direct access code: 220005-0xx
Examples: Universal input 1: 220005-000; Universal input 12: 220005-011

Description

Number of places after decimal point for the display.
Only visible if signal ≠ Switched off

Selection

None, One (X.Y), Two (X.YY), Three (X.YYY), Four (X.YYYY), Five (X.YYYYY)

Factory setting

One (X.Y)

Lower frequency

Navigation

Expert → Inputs → Universal inputs → Universal input x → Lower frequency
Direct access code: 220018-0xx
Examples: Universal input 1: 220018-000; Universal input 12: 220018-011

Description

Configure the lower frequency that corresponds to the start of the measuring range.
Only visible if signal = Frequency input

User entry

0 to 12500 (Hz)Factory setting

5.0 (Hz)

Range start
Endress+Hauser

111

Appendix

Ecograph T, RSG35

Navigation

Expert → Inputs → Universal inputs → Universal input x → Range start
Direct access code: 220006-0xx
Examples: Universal input 1: 220006-000; Universal input 12: 220006-011

Description

Transmitters convert the physical measured variable to standardized signals. Enter thestart of the measuring range here.
• The start and end of the measuring range may not be identical.
• The start of the measuring range can also be larger than the end (e.g. for deep
wells).
• The parameter can be defined independently of the number of decimal placesconfigured for the measured value as these are only taken into consideration forthe display.

User entry

Number (max. 8 digits)

Factory setting

0 (Depends on the input signal selected)

Upper frequency

Navigation

Expert → Inputs → Universal inputs → Universal input x → Upper frequency
Direct access code: 220019-0xx
Examples: Universal input 1: 220019-000; Universal input 12: 220019-011

Description

Configure the upper frequency that corresponds to the end of the measuring range.
Only visible if signal = Frequency input

User entry

0 to 12500 (Hz)Factory setting

1000.0 (Hz)Meas. range end

Navigation

Expert → Inputs → Universal inputs → Universal input x → Meas. range end
Direct access code: 220007-0xx
Examples: Universal input 1: 220007-000; Universal input 12: 220007-011

Description

Transmitters convert the physical measured variable to standardized signals. Enter the endof the measuring range here.
• The start and end of the measuring range may not be identical.
• The end of the measuring range can also be smaller than the start (e.g. for deep
wells).
• The parameter can be defined independently of the number of decimal placesconfigured for the measured value as these are only taken into consideration forthe display.

User entry

Number (max. 8 digits)

Factory setting

100 (Depends on the input signal selected)

112

Endress+Hauser

Ecograph T, RSG35

Appendix

Zoom start

Navigation

Description

Expert → Inputs → Universal inputs → Universal input x → Zoom start
Direct access code: 220011-0xx
Examples: Universal input 1: 220011-000; Universal input 12: 220011-011
If the whole value range is not used, you can configure the lower value of the required
section here. The zoom has no influence on the storage.
• The zoom can also be set outside the measuring range. The only restriction is thatthe start and end of the zoom may not be identical.
• If the signal or the range is changed, the zoom is corrected if it no longer fits themeasuring range.
• The zoom start can also be larger than the zoom end. The device will automaticallyrotate the values on the display.

User entry

Number (max. 8 digits)

Factory setting

0 (Depends on the input signal selected)

Zoom end

Navigation

Description

Expert → Inputs → Universal inputs → Universal input x → Zoom end
Direct access code: 220012-0xx
Examples: Universal input 1: 220012-000; Universal input 12: 220012-011
Like "Zoom start". Enter the upper value of the required range here.
• The zoom can also be set outside the measuring range. The only restriction is thatthe start and end of the zoom may not be identical.
• If the signal or the range is changed, the zoom is corrected if it no longer fits themeasuring range.
• The zoom end can also be smaller than the zoom start. The device willautomatically rotate the values on the display.

User entry

Number (max. 8 digits)

Factory setting

100 (Depends on the input signal selected)

Damping

Navigation

Expert → Inputs → Universal inputs → Universal input x → Damping
Direct access code: 220008-0xx
Examples: Universal input 1: 220008-000; Universal input 12: 220008-011

Description

The more unwanted interference there is on the measurement signal the higher the valuethat should be entered here. Result: Fast changes will be damped/suppressed.
Only visible if signal = Current, Voltage, Resistance thermometer or Thermocouple

Endress+Hauser

113

Appendix

Ecograph T, RSG35

User entry

0 to 999.9 s

Factory setting

Current, voltage: 0.0 s
Resistance thermometer, thermocouple: 0.2 s

Comparison point

Navigation

Expert → Inputs → Universal inputs → Universal input x → Comparison point
Direct access code: 220013-0xx
Examples: Universal input 1: 220013-000; Universal input 12: 220013-011

Description

Intern: Compensation of the voltage error by measuring the terminal temperature.
Extern: Compensation of the voltage error by using an external controlled comparison
measurement point.
Only visible if signal = Thermocouple

Selection

Internal, External

Factory setting

Internal

Comparison temp.

Navigation

Expert → Inputs → Universal inputs → Universal input x → Comparison temp.
Direct access code: 220014-0xx
Examples: Universal input 1: 220014-000; Universal input 12: 220014-011

Description

Settings for the external comparison temperature (only when connecting thermocouples).
Only visible if comparison point = External

User entry

0 to 9999999 (Depends on the temperature unit selected)

Factory setting

0 (Depends on the temperature unit selected)

Totalizer

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalizer
Direct access code: 220015-0xx
Examples: Universal input 1: 220015-000; Universal input 12: 220015-011

Description

Initial setting for the totalizer. Useful when continuing measurements recorded to datewith an (electro)-mechanical counter.
Only visible if signal = Pulse counter

User entry

Number (max. 15 digits)

Factory setting

0114

Endress+Hauser

Ecograph T, RSG35

Appendix"Meas.val. corrct." submenu
Navigation
Description

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct.
Determining the correction values to balance measurement tolerances.
Proceed as follows:• Measure the current value at the lower measurement range.
• Measure the current value at the upper measurement range.
• Enter the lower and upper target and actual value.

OffsetNavigation

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct. → Offset
Direct access code: 220050-0xx
Examples: Universal input 1: 220050-000; Universal input 12: 220050-011

Description

This offset is only effective on the analog input signal (no maths/bus channels).
Only visible if signal = Resistance thermometer or Thermocouple

User entry

Number (max. 8 digits)

Factory setting0

Correction RPT

Navigation

Description

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct. →Correction RPT
Direct access code: 220057-0xx
Examples: Universal input 1: 220057-000; Universal input 12: 220057-011
Rear panel temperature correction value for this analog input (only required forthermocouples).
Only visible/editable if the service code has been entered.

User entry

Number (max. 8 digits)

Factory setting-3.0 for slot 1+2-3.2 for slot 3

Range start

Endress+Hauser

115

Appendix

Ecograph T, RSG35

Target value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct. → Target
value
Direct access code: 220052-0xx
Examples: Universal input 1: 220052-000; Universal input 12: 220052-011

Description

Enter the lower set point here (e.g. measuring range 0°C to 100°C: 0°C).
Only visible if signal = Current or Voltage

User entry

Number (max. 8 digits)

Factory setting0

Actual value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct. → Actual
value
Direct access code: 220053-0xx
Examples: Universal input 1: 220053-000; Universal input 12: 220053-011

Description

Enter the lower value actually measured here (e.g. measuring range 0°C to 100°C:
measured value 0.5°C).
Only visible if signal = Current or Voltage

User entry

Number (max. 8 digits)

Factory setting0

Meas. range end

Target value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct. → Target
value
Direct access code: 220055-0xx
Examples: Universal input 1: 220055-000; Universal input 12: 220055-011

Description

Enter the upper set point here (e.g. measuring range 0°C to 100°C: 100°C).
Only visible if signal = Current or Voltage

User entry

Number (max. 8 digits)

Factory setting

100

116

Endress+Hauser

Ecograph T, RSG35

Appendix

Actual value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Meas.val. corrct. → Actual
value
Direct access code: 220056-0xx
Examples: Universal input 1: 220056-000; Universal input 12: 220056-011

Description

Enter the upper value actually measured here (e.g. measuring range 0°C to 100°C:
measured value 100.5°C).
Only visible if signal = Current or Voltage

User entry

Number (max. 8 digits)

Factory setting

100"Totalization" submenu
Navigation
Description

Expert → Inputs → Universal inputs → Universal input x → Totalization
Set up only required for totalization of flow or power consumption.

Totalization

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalization →Totalization
Direct access code: 220030-0xx
Examples: Universal input 1: 220030-000; Universal input 12: 220030-011

Description

By totalizing the analog signal (e.g. flow rate in m³/h) quantities (in m³) can be calculated.

Selection

No, Yes

Factory setting

No

Totalization base

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalization →Totalization baseDirect access code: 220031-0xx
Examples: Universal input 1: 220031-000; Universal input 12: 220031-011

Description

Select the required time base. Example: ml/s -> time base seconds (s); m³/h -> time base
hours (h).
Only visible if totalization = Yes

Selection

Second (s), Minute (min), Hour (h), Day (d)Endress+Hauser

117

Appendix

Factory setting

Ecograph T, RSG35

Second (s)Unit

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalization → Unit
Direct access code: 220032-0xx
Examples: Universal input 1: 220032-000; Universal input 12: 220032-011

Description

Enter the unit for the calculated quantity (e.g. "m³").
Only visible if totalization = Yes

User entry

Text (max. 6 characters)

Low flow cut off

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalization → Low flow
cut off
Direct access code: 220033-0xx
Examples: Universal input 1: 220033-000; Universal input 12: 220033-011

Description

If the volume flow recorded is below the set value, these quantities are not added to thecounter.
If the input is scaled from 0 to y, or if the pulse input is used, all values that are smaller
than the set value are not recorded.
If the input is scaled from -x to +y, all values around the zero point (e.g. also negativevalues) are not recorded.
Only visible if totalization = Yes

User entry

Number (max. 8 digits)

Factory setting0

Calc. factor

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalization → Calc.
factorDirect access code: 220034-0xx
Examples: Universal input 1: 220034-000; Universal input 12: 220034-011

Description

Factor for calculating the integrated value (e.g. the transmitter delivers l/s -> totalizationbase = second -> engineering unit required is m³ --> enter factor 0.001)
Only visible if totalization = Yes

User entry

Number (max. 8 digits)

Factory setting

1.0

118

Endress+Hauser

Ecograph T, RSG35

Appendix

Totalizer

Navigation

Expert → Inputs → Universal inputs → Universal input x → Totalization → Totalizer
Direct access code: 220035-0xx
Examples: Universal input 1: 220035-000; Universal input 12: 220035-011

Description

Initial setting for the totalizer. Useful when continuing measurements recorded to datewith an (electro)-mechanical counter.
Only visible if totalization = Yes

User entry

Number (max. 15 digits)

Factory setting0"Fault mode" submenu
In the event of an error the alarm relay is switched if so configured →  89
Navigation
Description

Expert → Inputs → Universal inputs → Universal input x → Fault mode
Contains settings that define how this channel is to react under fault conditions (e.g. cable
open circuit, over range).

NAMUR NE 43

Navigation

Expert → Inputs → Universal inputs → Universal input x → Fault mode → NAMUR
NE 43
Direct access code: 220060-0xx
Examples: Universal input 1: 220060-000; Universal input 12: 220060-011

Description

Activate/deactivate the 4-20mA loop monitoring as per NAMUR recommendation NE 43.
The following error ranges apply when NAMUR NE43 is switched on:≤ 3.8 mA: under range≥ 20.5 mA: over range≤ 3.6 mA or ≥ 21.0 mA: sensor error≤ 2 mA: cable open circuit
Only visible if signal = "Current" and range = "4-20 mA" or "4-20 mA squared".

Selection

Off, On

Factory setting

On

Cable open circuit

Endress+Hauser

119Appendix

Ecograph T, RSG35

Navigation

Expert → Inputs → Universal inputs → Universal input x → Fault mode → Cable open
circuitDirect access code: 220060-0xx
Examples: Universal input 1: 220060-000; Universal input 12: 220060-011

Description

Cable open circuit detection
Only visible if signal = "Voltage" and range = "1-5 V" or "1-5 V squared".

Selection

Off, On

Factory setting

On

Lower error value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Fault mode → Lower error
value
Direct access code: 220065-0xx
Examples: Universal input 1: 220065-000; Universal input 12: 220065-011

Description

When NE43 is switched off, defines the value that must be undershot for the device tooutput an error.
Only visible if signal = "Current", range = "4-20 mA" and NAMUR NE 43 = "Off"User entry

Number (max. 8 digits); 0 to 4 mA

Factory setting

3.9mA

Upper error value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Fault mode → Upper error
value
Direct access code: 220066-0xx
Examples: Universal input 1: 220066-000; Universal input 12: 220066-011

Description

When NE43 is switched off, defines the value that must be exceeded for the device tooutput an error.
Only visible if signal = "Current", range = "4-20 mA" and NAMUR NE 43 = "Off"User entry

Number (max. 8 digits); 20 to 22mA

Factory setting

20.8mATime delay

120

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Inputs → Universal inputs → Universal input x → Fault mode → Time delay
Direct access code: 220064-0xx
Examples: Universal input 1: 220064-000; Universal input 12: 220064-011

Description

The device only reacts (e.g. relay switch) when this condition, open circuit/under range/over range, has been operational for the preset time.
Only visible if NAMUR NE 43 = On

User entry

0 to 99 s

Factory setting

0s

On error

Navigation

Description

Expert → Inputs → Universal inputs → Universal input x → Fault mode → On error
Direct access code: 220061-0xx
Examples: Universal input 1: 220061-000; Universal input 12: 220061-011
Configure what value the device should continue working with (for calculations) if themeasured value is not valid (e.g. cable open circuit).
In the event of an error value, all the dependent calculations are flagged accordingly
as "error value". Counters are not flagged, however!

Selection

Invalid calculation, Error value

Factory setting

Invalid calculation

Error value

Navigation

Expert → Inputs → Universal inputs → Universal input x → Fault mode → Error value
Direct access code: 220062-0xx
Examples: Universal input 1: 220062-000; Universal input 12: 220062-011

Description

The device continues calculating with this value in the event of an error.
Only visible if on error = Error value

User entry

Number (max. 8 digits)

Factory setting0

Save event

Navigation

Endress+Hauser

Expert → Inputs → Universal inputs → Universal input x → Fault mode → Save event
Direct access code: 220063-0xx
Examples: Universal input 1: 220063-000; Universal input 12: 220063-011

121Appendix

Ecograph T, RSG35

Description

Stores a message in the event log when a fault occurs.

Selection

No, Yes

Factory settingNo

Copy settings

Navigation

Expert → Inputs → Universal inputs → Universal input x → Copy settings
Direct access code: 220200-0xx
Examples: Universal input 1: 220200-000; Universal input 12: 220200-011

Description

Copies settings from actual channel to selected channel.

Selection

Switched off, Universal input x
Users can choose from all the available universal inputs.

Factory setting

Switched off"Digital inputs -> Digital input x" submenu
Navigation
Description

Expert → Inputs → Digital inputs → Digital input x
Setting up only required if the digital inputs (e.g. events) are to be used.
x = place holder for selected digital input

Add input

Navigation

Expert → System → Digital inputs → Add input
Direct access code: 252000/000

Description

Addition of a digital input that must be configured according to the function.

Selection

No, Digital input x

Factory setting

No

Delete input

Navigation

Description
122

Expert → System → Digital inputs → Delete input
Direct access code: 252001/000
Delete an input configuration.
Endress+Hauser

Ecograph T, RSG35

Appendix

Selection

No, Digital input x

Factory setting

No

Function

Navigation

Expert → Inputs → Digital inputs → Digital input x → Function
Direct access code: 250000-00x
Examples: Digital input 1: 250000-000; Digital input 6: 250000-005

Description

Select the required function. Digital inputs are High active; this means the described effectis achieved by a high input.
Low = -3 to +5 V
High = +12 to +30 V

Selection

Switched off, Control input, On/off event, Pulse counter, Operational time, Event+operation time, Quantity from time, Modbus Slave (option)

Factory setting

Switched off

Function

Navigation

Expert → Inputs → Digital inputs → Digital input x → Function
Direct access code: 250014-00x
Examples: Digital input 1: 250014-000; Digital input 6: 250014-005

Description

Specifies how the data from the fieldbus is interpreted/processed.
Only visible if function = Modbus Slave

Selection

Switched off, Control input, On/off event, Pulse counter, Operational time, Event +operation time, Quantity from time

Factory setting

Switched off

Channel ident.

Navigation

Expert → Inputs → Digital inputs → Digital input x → Channel ident.
Direct access code: 250001-00x
Examples: Digital input 1: 250001-000; Digital input 6: 250001-005

Description

Measurement point name (e.g. "Pump") or description of the function of this input (e.g.
"Fault message").
Only visible if function ≠ Switched off

User entry

Text (max. 16 characters)

Endress+Hauser

123

Appendix

Factory setting

Ecograph T, RSG35

Digital x

Engineering unit

Navigation

Expert → Inputs → Digital inputs → Digital input x → Engineering unit
Direct access code: 250002-00x
Examples: Digital input 1: 250002-000; Digital input 6: 250002-005

Description

Technical unit of the count input, e.g. gal, cf, ... .
Only visible if function = Pulse counter or Quantity from time

User entry

Text (max. 6 characters)

Decimal point

Navigation

Expert → Inputs → Digital inputs → Digital input x → Decimal point
Direct access code: 250004-00x
Examples: Digital input 1: 250004-000; Digital input 6: 250004-005

Description

Number of places after decimal point for the display.
Only visible if function = Pulse counter or Quantity from time

Selection

None, One (X.Y), Two (X.YY), Three (X.YYY), Four (X.YYYY), Five (X.YYYYY)

Factory setting

One (X.Y)

Input factor in

Navigation

Expert → Inputs → Digital inputs → Digital input x → Input factor in
Direct access code: 250019-00x
Examples: Digital input 1: 250019-000; Digital input 6: 250019-005

Description

Defines if the set up factor is respective to 1 second or 1 hour.
Only visible if function = Quantity from time

Selection

Seconds, Hours

Factory setting

Seconds

Pulse value

124

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Inputs → Digital inputs → Digital input x → Pulse value
Direct access code: 250005-00x
Examples: Digital input 1: 250005-000; Digital input 6: 250005-005

Description

Factor, that when multiplied by the input signal results in the required physical value.
Examples:1 pulse equals 5 m3-> enter "5" here.
Only visible if function = Pulse counter

User entry

Number (max. 8 digits)

Factory setting

1

1 second=/1 hour= (depends on the setting in "Input factor in")Navigation

Expert → Inputs → Digital inputs → Digital input x → 1 second= / 1 hour=
Direct access code: 250005-00x
Examples: Digital input 1: 250005-000; Digital input 6: 250005-005

Description

Factor, that when multiplied by the operating time results in the required physical value.
Examples:1 second equals 8 l -> enter "8" here.
Only visible if function = Quantity from time

User entry

Number (max. 8 digits)

Factory setting

1

Time delay

Navigation

Expert → Inputs → Digital inputs → Digital input x → Time delay
Direct access code: 250017-00x
Examples: Digital input 1: 250017-000; Digital input 6: 250017-005

Description

The high signal must be active for at least the preset time before the device changes thechannel from low to high.
The change from high to low is always immediate.
Only visible if function = Control input, On/off event, Event+operation time

User entry

0 to 99 999 s

Factory setting0

ActionEndress+Hauser

125

Appendix

Ecograph T, RSG35

Navigation

Description

Expert → Inputs → Digital inputs → Digital input x → Action
Direct access code: 250003-00x
Examples: Digital input 1: 250003-000; Digital input 6: 250003-005
Set up the function of the control input.
Only visible if function = Control input
ActionDescription

Start/stop recording

The device only saves data as long as a high signal is present

Screensaver on

Switches backlighting/display off, low = off, high = on

Lock setup

The user can only change the setup if a low signal is present

Time synchronization

If a high signal is applied, the device rounds the system time up or down (only for
Low→ High change): 0 to 29 → round down; 30 to 59 → round up

Set point monitoring
on/off

The entire set point monitoring function of the device can be switched on (for"High") or switched off (for "Low").

Block keyboard/navigator

The device can only be operated if a low signal is present. Otherwise all key
activation and navigator actions are discarded.

Start/stop analysis 1

Starts/ends the external analysis (the analysis only runs as long as the signal ishigh). Measured value acquisition for the graphic display continues.

Selection

Switched off, Start/stop recording, Screensaver on, Lock setup, Time synchronization, Set
point monitoring on/off, Block keyboard/navigator, Start/stop analysis 1

Factory setting

Switched off

Switches relay

Navigation

Expert → Inputs → Digital inputs → Digital input x → Switches relay
Direct access code: 250006-00x
Examples: Digital input 1: 250006-000; Digital input 6: 250006-005

Description

Switches the respective relay when the digital input is low or high. Please take note of theconnection hints in the operating manual!
Only visible if function = Control input, On/off event, Event+operation time

Selection

Not used, Relay x
All the available relays are displayed.

Factory setting

Not used

Description 'H'

Navigation

126

Expert → Inputs → Digital inputs → Digital input x → Description 'H'
Direct access code: 250007-00x
Examples: Digital input 1: 250007-000; Digital input 6: 250007-005

Endress+Hauser

Ecograph T, RSG35

Appendix

Description

Condition description when the digital input is active. This text is both shown on thedisplay and saved to memory.
Only visible if function = Control input, On/off event, Event+operation time

User entry

Text (max. 6 characters)

Factory settingOn

Description 'L'

Navigation

Expert → Inputs → Digital inputs → Digital input x → Description 'L'
Direct access code: 250008-00x
Examples: Digital input 1: 250008-000; Digital input 6: 250008-005

Description

Condition description when the digital input is not active. This text is both shown on thedisplay and saved to memory.
Only visible if function = Control input, On/off event, Event+operation time

User entry

Text (max. 6 characters)

Factory setting

Off

Save event

Navigation

Description

Expert → Inputs → Digital inputs → Digital input x → Save event
Direct access code: 250009-00x
Examples: Digital input 1: 250009-000; Digital input 6: 250009-005
Determines whether the condition change from low to high or high to low is stored in theevent log.
Requires higher memory capacity.
Only visible if function = Control input, On/off event, Event+operation time

Selection

No, Yes, only "On" message

Factory setting

Yes

Event message

Navigation

Endress+Hauser

Expert → Inputs → Digital inputs → Digital input x → Event message
Direct access code: 250018-00x
Examples: Digital input 1: 250018-000; Digital input 6: 250018-005

127Appendix

Ecograph T, RSG35

Description"Do not acknowledge": No message is shown if the digital input switches.
"Acknowledge": A message window is shown on the screen which has to be acknowledgedby operating a push button.
Only visible if function = Control input, On/off event, Event+operation time

Selection

Do not acknowledge, Acknowledge

Factory setting

Do not acknowledge

Event text L->H

Navigation

Description

Expert → Inputs → Digital inputs → Digital input x → Event text L->H
Direct access code: 250010-00x
Examples: Digital input 1: 250010-000; Digital input 6: 250010-005
Description of condition change from low to high. Event text is stored (e.g. Start filling).
If no event text is set, the device generates an automatic event text (factory setting),
e.g. digital 1 L->H.
Only visible if function = Control input, On/off event, Event+operation time

User entry

Text (max. 22 characters)

Event text H->L

Navigation

Description

Expert → Inputs → Digital inputs → Digital input x → Event text H->L
Direct access code: 250011-00x
Examples: Digital input 1: 250011-000; Digital input 6: 250011-005
Description of condition change from high to low. Event text is stored (e.g. Stop filling).
If no event text is set, the device generates an automatic event text (factory setting),
e.g. digital 1 H->L.
Only visible if function = Control input, On/off event, Event+operation time

User entry

Text (max. 22 characters)

Record duration

Navigation

128

Expert → Inputs → Digital inputs → Digital input x → Record duration
Direct access code: 250012-00x
Examples: Digital input 1: 250012-000; Digital input 6: 250012-005

Endress+Hauser

Ecograph T, RSG35

Appendix

Description

The duration between "On" and "Off" can be recorded. The duration is appended to the "Off"event text (<hhhh>h<mm>:<ss>).
Power failure times do not affect the duration. If the digital channel was "on" before thepower failure and is still "on" after the power failure, the duration continues.
Only visible if function = Control input, On/off event, Event+operation time

Selection

No, Yes

Factory setting

No

Totalizer

Navigation

Expert → Inputs → Digital inputs → Digital input x → Totalizer
Direct access code: 250013-00x
Examples: Digital input 1: 250013-000; Digital input 6: 250013-005

Description

Initial setting for the totalizer. Useful when continuing measurements recorded to datewith an (electro)-mechanical counter.
Only visible if function = Pulse counter, Operational time, Event+operation time or
Quantity from time

User entry

Number (max. 15 digits)

Factory setting0

Copy settings

Navigation

Expert → Inputs → Digital inputs → Digital input x → Copy settings
Direct access code: 250200-00x
Examples: Digital input 1: 250200-000; Digital input 6: 250200-005

Description

Copies settings from actual channel to selected channel.

Selection

No, Digital input x
Users can choose from all the available digital inputs.

Factory settingNo

16.1.3"Outputs" submenu

Setting up only required if outputs (e.g. relays) are to be used.

"Relay x" submenu
Navigation

Endress+Hauser

Expert → Outputs → Relay x

129

Appendix

Description

Ecograph T, RSG35

Contains setup for the selected relay
x = place holder for selected relay

Operating mode

Navigation

Expert → Outputs → Relay x → Operating mode
Direct access code: 330000-00x
Examples: Relay 1:330000-000; Relay 6: 330000-005

Description

Relay function:
NC contact: The relay is closed in its quiescent state (maximum safety).
NO contact: The relay is open in its quiescent state.

Selection

NO contact, NC contact

Factory setting

NO contact

Identifier

Navigation

Expert → Outputs → Relay x → Identifier
Direct access code: 330001-00x
Examples: Relay 1:330001-000; Relay 6: 330001-005

Description

Presettable relay identifier.

User entry

Text (max. 16 characters)

Factory setting

Relay x16.1.4"Communication" submenu

Set-up required if you are using the USB, RS232, RS485 or Ethernet interface of the unit
(PC operation, serial data read-out, modem operation, etc.).
The various interfaces can be operated in parallel.

Timeout cycl. readout

Navigation

Expert → Communication → Timeout
Direct access code: 150200-000

Description

Monitors whether measured values are read out cyclically via OPC or Fieldbus. Timeout is
modifiable between 1 and 99 seconds. 0 seconds means that the functionality isinactivated.

User entry

0 to 99 s

130

Endress+Hauser

Ecograph T, RSG35

Factory setting

Appendix0s

Switches

Navigation

Expert → Communication → Switches
Direct access code: 150201-000

Description

After the given timeout the dedicated relay/OC is active while no readout of actual
measured values are in process.

Selection

Not used, Relay x
All the available relays are displayed.

Factory setting

Not used

Timeout fieldbus

Navigation

Expert → Communication → Timeout fieldbus
Direct access code: 150210-000

Description

Time within which measured values must be received via fieldbus (otherwise an error will
be set). Not relevant if only measured values are read out.

User entry

1 to 99s

Factory setting

10 s

Function USB-B

Navigation

Expert → Communication → Function USB-B
Direct access code: 012001-000

Description

Determines the operating mode of the USB interface if a cable is connected to the device.

Selection

Always USB
Always Ethernet over USB
By user input

Factory setting

Always USB"Ethernet" submenu
Navigation
Endress+Hauser

Expert → Communication → Ethernet
131

Appendix

Description

Ecograph T, RSG35

Contains the set-up required if you are using the Ethernet interface of the unit.

MAC address

Navigation

Description

Expert → Communication → Ethernet → MAC address
Direct access code: 150000-000
Displays the MAC address

DHCP

Navigation

Description

Expert → Communication → Ethernet → DHCP
Direct access code: 150002-000
The device can get its Ethernet settings through DHCP.
Caution: The settings determined are not displayed until after setup acceptance!
Note: The unit always gets the same IP address if the leasing time is set long enoughon the DHCP server. The PC software needs the IP address determined to establish aconnection!

Selection

No, Yes

Factory setting

Yes

IP address

Navigation

Expert → Communication → Ethernet → IP address
Direct access code: 150003-000

Description

Please enter the IP address. (given from your network administrator).
Can only be edited if DHCP = No

User entry

IP address

Factory setting

000.000.000.000

Subnetmask

Navigation

Description

132

Expert → Communication → Ethernet → Subnetmask
Direct access code: 150004-000
Please enter the subnetmask (given from your network administrator).
Can only be edited if DHCP = No
Endress+Hauser

Ecograph T, RSG35

Appendix

User entry

IP address

Factory setting

255.255.255.000

Gateway

Navigation

Expert → Communication → Ethernet → Gateway
Direct access code: 150005-000

Description

Please enter the gateway (given from your network administrator).
Can only be edited if DHCP = No

User entry

IP address

Factory setting

000.000.000.000

Domain Name System (DNS)

Navigation

Expert → Communication → Ethernet → Domain Name System (DNS)
Direct access code: 150009-000

Description

Please enter the IP-address of the DNS server (you can get this from your network
administrator).
Is needed if you wish to send e-mails and want to use the e-mail server name instead of
the IP address (e.g. smtp.example.org).
Can only be edited if DHCP = No

User entry

IP address

Factory setting

000.000.000.000

Disable port

Navigation

Description

Expert → Communication → Ethernet → Disable port
Direct access code: 150020-000
You can disable unused ports for security reasons.
CDI is the protocol that the configuration software or reporting software uses tocommunicate with the device.
All other ports (e.g. SNTP, SMTP, Web server) are automatically disabled if thefunction is switched off.

Selection

CDI, OPC, Modbus Slave

Factory setting---- (no port disabled)

Endress+Hauser

133

Appendix

Ecograph T, RSG35

Port

Navigation

Description

Expert → Communication → Ethernet → Port
Direct access code: 150001-000
The system communicates with the PC software through this communication port.
If your network is protected by a firewall, this port may have to be enabled. Pleasecontact your network administrator if this is the case.

User entry

Number (max. 5 digits)

Factory setting

8000

OPC port

Navigation

Expert → Communication → Ethernet → OPC port
Direct access code: 150010-000

Description

Values van be read via OPC server using this communication port.
If your network is protected by a firewall, this port may have to be enabled. Pleasecontact your network administrator if this is the case.

User entry

Number (max. 5 digits)

Factory setting

8002

Web server

Navigation

Description

Expert → Communication → Ethernet → Web server
Direct access code: 470000-000
Switch the Web server function on or off. The instantaneous values can only be displayedusing an Internet browser when the web browser is activated.
It is only possible to connect to the Web server via the Ethernet interface!

Selection

No (Web server is off), Yes (Web server is active)

Factory setting

Yes"Configuration Web server" submenu
Navigation
134

Expert → Communication → Ethernet → Configuration Web serverEndress+Hauser

Ecograph T, RSG35

Description

Appendix

Configure the Web server or specify which functionality should be possible via Web server.
Only visible if Web server = Yes.
Instantaneous value display is always possible once the Web server is switched on.

Port

Navigation

Description

Expert → Communication → Ethernet → Configuration Web server → Port
Direct access code: 470003-000
The Web server communicates through this communication port.
If your network is protected by a firewall, this port may have to be enabled.
Please contact your network administrator if this is the case.

User entry

Number (max. 5 digits)

Factory setting

80

Setup

Navigation

Expert → Communication → Ethernet → Configuration Web server → Setup
Direct access code: 470001-000

Description

The device can be configured via Web server.
For security reasons it is advisable to switch off configuration via the Web server aftercommissioning.
With regard to IT security please contact your network administrator if necessary.

Selection

No, Yes

Factory setting

Yes

Firmware update

Navigation

Expert → Communication → Ethernet → Configuration Web server → Firmware
updateDirect access code: 470002-000

Description

Firmware can be updated via Web server.

Selection

No, Yes

Factory setting

No

Endress+Hauser

135Appendix

Ecograph T, RSG35

Remote control

Navigation

Expert → Communication → Ethernet → Configuration Web server → Remotecontrol
Direct access code: 470004-000

Description

The device can be remote-controlled via the Web server.

Selection

No, Yes

Factory setting

No

WebDAV server

Navigation

Expert → Communication → Ethernet → Configuration Web server → WebDAV
serverDirect access code: 470006-000

Description

The SD card can be read out via the WebDAV client.

Selection

No, Yes

Factory settingNo

Meas. val. without login

Navigation

Expert → Communication → Ethernet → Configuration Web server → Meas. val.
without login
Direct access code: 470009-000

Description

Allow access to current measured values without logging in.
URL: http:\\<ip>\iv

Selection

Yes, No

Factory setting

Yes"Authentication" submenu
Navigation
Description

136

Expert → Communication → Ethernet → Configuration Web server → Authentication
Set the passwords for the various users with which the device can be accessed via Web
server.

Endress+Hauser

Ecograph T, RSG35

Appendix

Operator

Admin

Service

Measured value display

Yes

Yes

Yes

Display instrument health status

Yes

Yes

Yes

Configuration

No

Yes

Yes

Configuration incl. service parameter

No

No

Yes

Update firmware

No

Yes

Yes

WebDAV

Yes

Yes

Yes

Note: The following passwords should be changed during commissioning.

Operator

ID

Navigation

Expert → Communication → Ethernet → Configuration Web server → Authentication→ ID
Direct access code: 470104-000

Description

ID required in order to access the device. Pay attention to case-sensitivity.
Cannot be edited.

Factory setting

operator

Password

Navigation

Expert → Communication → Ethernet → Configuration Web server → Authentication→ Password
Direct access code: 470105-000

Description

Enter a password for this user account.
Pay attention to case-sensitivity.

User entry

Text (max. 12 characters)

Factory setting

operator

Administrator

ID

Endress+Hauser

137

Appendix

Ecograph T, RSG35

Navigation

Expert → Communication → Ethernet → Configuration Web server → Authentication→ ID
Direct access code: 470101-000

Description

ID required in order to access the device. Pay attention to case-sensitivity.
Cannot be edited.

Factory setting

admin

Password

Navigation

Expert → Communication → Ethernet → Configuration Web server → Authentication→ Password
Direct access code: 470102-000

Description

Enter a password for this user account.
Pay attention to case-sensitivity.

User entry

Text (max. 12 characters)

Factory setting

adminServiceID

Navigation

Expert → Communication → Ethernet → Configuration Web server → Authentication→ ID
Direct access code: 470107-000

Description

ID required in order to access the device. Pay attention to case-sensitivity.
Cannot be edited.

Factory setting

servicePassword

Navigation

Description

138

Expert → Communication → Ethernet → Configuration Web server → Authentication→ Password
Direct access code: 470108-000
Enter a password for this user account.
Pay attention to case-sensitivity.

Endress+Hauser

Ecograph T, RSG35

Appendix

User entry

Text (max. 12 characters)

Factory setting

service"Timeouts" submenu
Navigation

Expert → Communication → Ethernet → Configuration Web server → Timeouts

Description

Timeouts for the Web server. Settings should only be modified if slow network connections
are causing transmission problems.
The settings are only adopted if the browser has been restarted or a new tab opened.
Caution: Settings should only be changed by experts.

Connection quality

Navigation

Description

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Connection quality
Direct access code: 470200-000
Configuration of typical timeout values for the Web server connection.
The default values can be changed if necessary.

Selection

Please select, Local network (LAN/WLAN), Wireless/mobile (fast connect.), Wireless/mobile (slow connect)

Factory setting

Please selectGet timeout

Navigation

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Get timeout
Direct access code: 470201-000

Description

Maximum time to load a new page before the browser ends the connection.

User entry

5 to 999 s

Factory setting

25

Set timeout

Endress+Hauser

139

Appendix

Ecograph T, RSG35

Navigation

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Set timeout
Direct access code: 470202-000

Description

Maximum time to write a value or execute an action before the browser ends theconnection.

User entry

5 to 999 s

Factory setting

5

Put timeout

Navigation

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Put timeout
Direct access code: 470203-000

Description

Maximum time to transmit files to or from the device before the browser ends theconnection.

User entry

5 to 9999 s

Factory setting

240

Ping interval

Navigation

Description

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Ping interval
Direct access code: 470204-000
Interval in which the browser checks device reachability.
The check is switched off if 0s is set. This is for diagnostic purposes only and should
not be set!

User entry

0 to 999 s

Factory setting

10

Ping timeout

Navigation

Description

140

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Ping timeout
Direct access code: 470205-000Time in which the device must respond before the browser ends the connection.

Endress+Hauser

Ecograph T, RSG35

Appendix

User entry

5 to 999 s

Factory setting

15

Ping retry

Navigation

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Ping retry
Direct access code: 470206-000

Description

Number of retries if the device does not respond.

User entry

0 to 5

Factory setting0Poll timeout

Navigation

Expert → Communication → Ethernet → Configuration Web server → Timeouts →Poll timeout
Direct access code: 470207-000

Description

Maximum permitted time to refresh the website.

User entry

5 to 999 s

Factory setting

5"Serial interface" submenu
Navigation
Description

Expert → Communication → Serial interface
Contains the set-up required if you are using the RS232 or RS485 unit interface.

Type

Navigation

Expert → Communication → Serial interface → Type
Direct access code: 150100-000

Description

Configure how the serial interface is used. Pay attention to the connections.

Selection

RS232, RS485, Debug (only for service purposes)

Factory setting

RS232

Endress+Hauser

141

Appendix

Ecograph T, RSG35

ProtocolNavigation

Expert → Communication → Serial interface → Protocol
Direct access code: 150105-000

Description

Define the serial interface protocol.
Note: The device automatically disables incompatible settings.

Selection

PC software, Modbus Slave (only if type = RS485)

Factory setting

PC software

Baud rate

Navigation

Expert → Communication → Serial interface → Baud rate
Direct access code: 150101-000

Description

Transmission speed ("Baud rate") - must be the same as the settings for the PC software.

Selection

9600, 19200, 38400, 57600, 115200

Factory setting

19200

ParityNavigation

Expert → Communication → Serial interface → Parity
Direct access code: 150103-000

Description

ParityOnly visible if protocol ≠ PC software

Selection

None, Even, Odd

Factory setting

None

Stop bits

Navigation

Description

142

Expert → Communication → Serial interface → Stop bits
Direct access code: 150104-000
Stop bits
Only visible if protocol ≠ PC software
Endress+Hauser

Ecograph T, RSG35

Appendix

Selection

1, 2Factory setting

1

Device address

Navigation

Expert → Communication → Serial interface → Device address
Direct access code: 150102-000

Description

Every device operating using RS232 / RS485 must have an individual address (00-30).
Only visible if type = RS485

User entry

0 to 30

Factory setting

0"Modbus Slave" submenu (option)
Navigation
Description

Expert → Communication → Modbus Slave
Configure the Modbus settings for the device.
Detailed descriptions of this device option can be found in the associateddocumentation.

ModbusNavigation

Expert → Communication → Modbus Slave → Modbus
Direct access code: 480000-000

Description

Specify the physical interface you wish to use.

Selection

Not used, RS485, Ethernet

Factory setting

Not used

Device address

Navigation

Expert → Communication → Modbus Slave → Device address
Direct access code: 480001-000

Description

Enter the device address where it should be possible to reach this device in the bus.
Only visible if Modbus = RS485

User entry

1 to 247

Endress+Hauser

143

Appendix

Factory setting

Ecograph T, RSG35

1

Port

Navigation

Expert → Communication → Modbus Slave → Port
Direct access code: 480004-000

Description

Port via which the Modbus protocol can be activated.
Only visible if Modbus = Ethernet

User entry

Number (max. 5 digits)

Factory setting

502"Serial interface" submenu
Navigation
Description

Expert → Communication → Modbus Slave → Serial interface
Contains settings for the serial interface.
Only visible if Modbus = RS485

Baud rate

Navigation

Expert → Communication → Modbus Slave → Serial interface → Baud rate
Direct access code: 150101-000

Description

Transmission speed ("Baud rate") - must be the same as the settings for the PC software.
Only visible if Modbus = RS485

Selection

9600, 19200, 38400, 57600, 115200

Factory setting

19200

ParityNavigation

Expert → Communication → Modbus Slave → Serial interface → Parity
Direct access code: 150103-000

Description

ParityOnly visible if Modbus = RS485

Selection

None, Even, Odd

Factory setting

None

144

Endress+Hauser

Ecograph T, RSG35

Appendix

Stop bits

Navigation

Expert → Communication → Modbus Slave → Serial interface → Stop bits
Direct access code: 150104-000

Description

ParityOnly visible if Modbus = RS485 and parity = None

Selection

1, 2Factory setting

1

16.1.5"Application" submenu

Configure various application-specific settings (e.g. group settings, limit values, etc.).

"Maths - Maths x" submenu
Navigation
Description

Expert → Application → Maths → Maths x
Configuration of the mathematics channels.
x = place holder for selected mathematics channel

Function

Navigation

Expert → Application → Maths → Maths x → Function
Direct access code: 400000-000
Examples: Maths 1: 400000-000; Maths 4: 400000-003

Description

Switch the mathematics channel on or off.

Selection

Switched off, Formula editor

Factory setting

Switched off

Channel ident.

Navigation

Endress+Hauser

Expert → Application → Maths → Maths x → Channel ident.
Direct access code: 400001-000
Examples: Maths 1: 400001-000; Maths 4: 400001-003

145Appendix

Ecograph T, RSG35

Description

Measurement point name (e.g. "Pump") or description of the function of this input (e.g.
"Fault message").

User entry

Text (max. 16 characters)

Factory setting

Maths xFormulaNavigation

Expert → Application → Maths → Maths x → Formula
Direct access code: 400002-000
Examples: Maths 1: 400002-000; Maths 4: 400002-003

Description

Enter the desired calculation formula.
Analog, digital or already active mathematics channels can be used.
Description of formula editor →  151
Only visible if function = Formula editor

User entry

FormulaThe result is

Navigation

Expert → Application → Maths → Maths x → The result is
Direct access code: 400003-000
Examples: Maths 1: 400003-000; Maths 4: 400003-003

Description

Configure which data type the calculation returns. This setting affects how the channel
saves and is displayed.
If you add 2 analog channels, for example, the result is an "instantaneous value".
Instantaneous value: If, for example, 2 analog channels are added (AI(1;1)+AI(1;2)), theresult is an instantaneous value.
State: The state/status of an individual analog input can be output as the result. A relay
can also be actuated as a result.
Counter: If, for example, 2 counters from digital inputs are added (DI(3;1)+DI(3;5)), theresult is a counter.
Operating time from status: The status (logical "1" or "0") of one or more digital inputs thatare connected by addition can be analyzed. If the result of the calculation is not equal to 0,
the counter for the operating time starts. The time is increased by 0.1 s every 100 ms.
Operating time from total: If several digital inputs that are configured as "operational time"
are added together, the result is the total of all the individual operating times.
Control input: The function corresponds to a digital input that has been configured as acontrol input.

Selection

Instantaneous value, State, Counter, Operating time from status, Operating time fromtotal, Control input

Factory setting

Instantaneous value

146

Endress+Hauser

Ecograph T, RSG35

Appendix

Plot type

Navigation

Expert → Application → Maths → Maths x → Plot type
Direct access code: 400015-000
Examples: Maths 1: 400015-000; Maths 4: 400015-003

Description

The mathematics channels are recalculated every 100 ms.
Depending on the save cycle, the selected data are determined/saved from the calculated
values.

Selection

Instantaneous value, Average, Minimum value, Maximum value, Minimum + Maximum,
Counter, Current value + Counter

Factory setting

AverageEngineering unit

Navigation

Expert → Application → Maths → Maths x → Engineering unit
Direct access code: 400004-000
Examples: Maths 1: 400004-000; Maths 4: 400004-003

Description

Unit of the calculated value.
Only visible if the result is = Instantaneous value or Counter

User entry

Text (max. 6 characters)

Decimal point

Navigation

Expert → Application → Maths → Maths x → Decimal point
Direct access code: 400005-000
Examples: Maths 1: 400005-000; Maths 4: 400005-003

Description

Number of places after decimal point for the display.
Only visible if function = Formula editor and the result is = Instantaneous value or Counter

Selection

None, One (X.Y), Two (X.YY), Three (X.YYY), Four (X.YYYY), Five (X.YYYYY)

Factory setting

One (X.Y)

ActionNavigation

Endress+Hauser

Expert → Application → Maths → Maths x → Action
Direct access code: 400006-000
Examples: Maths 1: 400006-000; Maths 4: 400006-003
147

Appendix

Description

Ecograph T, RSG35

Set up the function of the control input.
Only visible if the result is = Control input
ActionDescription

Start/stop recording

The device only saves data as long as a high signal is present

Screensaver on

Switches backlighting/display off, low = off, high = on

Lock setup

The user can only change the setup if a low signal is present

Time synchronization

If a high signal is applied, the device rounds the system time up or down (only for
Low→ High change): 0 to 29 → round down; 30 to 59 → round up

Set point monitoring
on/off

The entire set point monitoring function of the device can be switched on (for"High") or switched off (for "Low").

Block keyboard/navigator

The device can only be operated if a low signal is present. Otherwise all key
activation and navigator actions are discarded.

Start/stop analysis 1

Starts/ends the external analysis (the analysis only runs as long as the signal ishigh). Measured value acquisition for the graphic display continues.

Selection

Switched off, Start/stop recording, Screensaver on, Lock setup, Time synchronization, Set
point monitoring on/off, Block keyboard/navigator, Start/stop analysis 1

Factory setting

Switched off

Switches relay

Navigation

Expert → Application → Maths → Maths x → Switches relay
Direct access code: 400007-000
Examples: Maths 1: 400007-000; Maths 4: 400007-003

Description

Switches the respective relay when the digital input is low or high.
Only visible if the result is = Control input or State

Selection

Not used, Relay x
All the available relays are displayed.

Factory setting

Not used

Description 'H'

Navigation

Expert → Application → Maths → Maths x → Description 'H'
Direct access code: 400008-00x
Examples: Maths 1: 400008-000; Maths 4: 400008-003

Description

Condition description when the digital input is active. This text is both shown on thedisplay and saved to memory.
Only visible if the result is = Control input or State

User entry

Text (max. 6 characters)

148

Endress+Hauser

Ecograph T, RSG35

Factory setting

AppendixOn

Description 'L'

Navigation

Expert → Application → Maths → Maths x → Description 'L'
Direct access code: 400009-00x
Examples: Maths 1: 400009-000; Maths 4: 400009-003

Description

Condition description when the digital input is not active. This text is both shown on thedisplay and saved to memory.
Only visible if the result is = Control input or State

User entry

Text (max. 6 characters)

Factory setting

Off

Save event

Navigation

Description

Expert → Application → Maths → Maths x → Save event
Direct access code: 400010-00x
Examples: Maths 1: 400010-000; Maths 4: 400010-003
Determines whether the condition change from low to high or high to low is stored in theevent log.
Requires higher memory capacity.
Only visible if the result is = Control input or State

Selection

No, Yes, only "On" message

Factory setting

Yes

Event message

Navigation

Expert → Application → Maths → Maths x → Event message
Direct access code: 400018-00x
Examples: Maths 1: 400018-000; Maths 4: 400018-003

Description"Do not acknowledge": No message is shown if the state of the mathematics channelchanges.
"Acknowledge": A message window is shown on the screen which has to be acknowledgedby operating a push button.
Only visible if the result is = Control input or State

Selection

Do not acknowledge, Acknowledge

Endress+Hauser

149

Appendix

Factory setting

Ecograph T, RSG35

Do not acknowledge

Event text L->H

Navigation

Expert → Application → Maths → Maths x → Event text L->H
Direct access code: 400011-00x
Examples: Maths 1: 400011-000; Maths 4: 400011-003

Description

Description of condition change from low to high. Event text is stored (e.g. Start filling).
Only visible if the result is = Control input or State

User entry

Text (max. 22 characters)

Event text H->L

Navigation

Expert → Application → Maths → Maths x → Event text H->L
Direct access code: 400012-00x
Examples: Maths 1: 400012-000; Maths 4: 400012-003

Description

Description of condition change from high to low. Event text is stored (e.g. Stop filling).
Only visible if the result is = Control input or State

User entry

Text (max. 22 characters)

Record duration

Navigation

Expert → Application → Maths → Maths x → Record duration
Direct access code: 400013-00x
Examples: Maths 1: 400013-000; Maths 4: 400013-003

Description

The duration between "On" and "Off" can be recorded. The duration is appended to the "Off"event text (<hhhh>h<mm>:<ss>).
Power failure times do not affect the duration. If the digital channel was "on" before thepower failure and is still "on" after the power failure, the duration continues.
Only visible if the result is = Control input or State

Selection

No, Yes

Factory settingNo

Zoom start

150

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Application → Maths → Maths x → Zoom start
Direct access code: 400016-00x
Examples: Maths 1: 400016-000; Maths 4: 400016-003

Description

If the whole value range is not used, you can configure the lower value of the required
section here. The zoom has no influence on the storage.
Only visible if the result is = Instantaneous value

User entry

Number (max. 8 digits)

Factory setting0

Zoom end

Navigation

Expert → Application → Maths → Maths x → Zoom end
Direct access code: 400017-00x
Examples: Maths 1: 400017-000; Maths 4: 400017-003

Description

Like "Zoom start". Enter the upper value of the required range here.
Only visible if the result is = Instantaneous value

User entry

Number (max. 8 digits)

Factory setting

100

Totalizer

Navigation

Expert → Application → Maths → Maths x → Totalizer
Direct access code: 400014-00x
Examples: Maths 1: 400014-000; Maths 4: 400014-003

Description

Initial setting for the totalizer. Useful when continuing measurements recorded to datewith an (electro)-mechanical counter.
Only visible if the result is = Counter, Operating time from status or Operating time from
total

User entry

Number (max. 15 digits)

Factory setting0

Formula editorEnter the desired calculation formula.
Analog, digital or already active mathematics channels can be used.

Formula editor

Endress+Hauser

151

Appendix

Ecograph T, RSG35

Navigation

Expert → Application → Maths → Maths x → Formula
Direct access code: 400002-000
x = place holder for selected mathematics channel
A text field with the formula currently used appears. If the field is empty a formula
has not yet been defined for the mathematics channel.

Description

Individual channels can be mathematically linked and calculated with functions. The
mathematics channels calculated in this way are treated as "real" channels, regardless ofwhether they are connected conventionally or via fieldbus. Enter the desired calculation
formula.
Analog and digital channels can be used, as can mathematics channels that are alreadyactive.
A formula with up to 200 characters can be created using this editor. If the formula isfinished, click OK to close the editor and accept the formula entered. The common entry
and arithmetic operators and inputs are described in detail in the following sections.
Inputs
Inputs are described in the formula using the following syntax:
Input type (signal type;channel number)Type of input

Description

AI

Analog inputs

DI

Digital inputs

MI

Mathematics inputs

Signal type

Description

1

Instantaneous value (measured value)

2

State

3

Counter/operational time5

Validity:
The validity of an analog channel or a mathematics channel is relayed.
The relayed value of the function is 0 in the event of:• Open circuit• Invalid measured value• Sensor error• Input signal too high/low• Error value
The relayed value of the function is 1 in the event of:
Measured value OK, even if the limit value is breached

6

Delta count7 to 10Analysis 1 to 4

11

Totalizer

12

Duration

Not all signal types are available for each input type. These depend on the respective
device options.
Channel number:
Analog channel 1 = 1, analog channel 2 = 2, digital channel 1 = 1, …

152

Endress+Hauser

Ecograph T, RSG35

Appendix

Examples:DI(2;4)

State of digital channel 4AI(1;1)

Instantaneous value of analog channel 1

Status of a limit value:
LMT (type, limit number)Type

Description

1"Instantaneous value": Currently set limit value

2"State": The function returns the status of a limit value
The result is 1 if the limit value is violated.
The result is 0 if
• The limit value is not violated• The limit value is not switched on• Limits monitoring is switched off (e.g. per control input)

Examples:LMT (1;1)

Instantaneous value of limit value 1

LMT (2;3)State of limit value 3

Priority of operators / functions
The formula is processed based on universally applicable mathematics rules:• Parentheses first• Exponents before multiplication or division• Multiplication or division before addition or subtraction• Calculate from left to rightOperators
Arithmetic operators:
Operator

Function+

Addition-

Subtraction / negative sign*

Multiplication/

Division

Decimal separator
Both the decimal point and the decimal comma can be used in the formula editor.
Thousand separators are not supported.
Check whether formula is valid or malfunctions
A formula is invalid if:• The channels used are not switched on or are in the wrong operating mode (is notverified during formula entry as the channel could be switched on subsequently)• It contains invalid characters/formulas/functions/operators• Syntax errors (e.g. wrong number of parameters) occur in the formulas

Endress+Hauser

153

Appendix

Ecograph T, RSG35• There are incorrect parentheses in the formula (number of open parentheses unequal tonumber of closed parentheses)• Division is by zero
• A channel refers to itself (infinite recursion)
Invalid formulas are deactivated when the setup is accepted or the device is started.
Undetectable errors: wherever possible, errors in the formula are reported immediately
during input. However, given the possible complexity of the formula entered (e.g. nested
formulas) it is not possible to detect every error.

"Totalization" submenu
Navigation
Description

Expert → Application → Maths → Maths x → Totalization
Settings only needed if the calculated value - e.g. for quantity calculation - should be
integrated. Analysis time frames, see "Signal analysis".

Totalization

Navigation

Expert → Application → Maths → Maths x → Totalization → Totalization
Direct access code: 400050-00x
Examples: Maths 1: 400050-000; Maths 4: 400050-003

Description

By totalizing the analog signal (e.g. flow rate in m³/h) quantities (in m³) can be calculated.

Selection

No, Yes

Factory setting

No

Totalization base

Navigation

Expert → Application → Maths → Maths x → Totalization → Totalization baseDirect access code: 400051-00x
Examples: Maths 1: 400051-000; Maths 4: 400051-003

Description

Select the required time base. Example: ml/s -> time base seconds (s); m³/h -> time base
hours (h).
Only visible if totalization = Yes

Selection

Second (s), Minute (min), Hour (h), Day (d)Factory setting

Second (s)

Unit

154

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Application → Maths → Maths x → Totalization → Unit
Direct access code: 400052-00x
Examples: Maths 1: 400052-000; Maths 4: 400052-003

Description

Enter the unit for the calculated quantity (e.g. "m³").
Only visible if totalization = Yes

User entry

Text (max. 6 characters)

Low flow cut off

Navigation

Expert → Application → Maths → Maths x → Totalization → Low flow cut off
Direct access code: 400053-00x
Examples: Maths 1: 400053-000; Maths 4: 400053-003

Description

If the volume flow recorded is below the set value, these quantities are not added to thecounter.
If the input is scaled from 0 to y, or if the pulse input is used, all values that are smaller
than the set value are not recorded.
If the input is scaled from -x to +y, all values around the zero point (e.g. also negativevalues) are not recorded.
Only visible if totalization = Yes

User entry

Number (max. 8 digits)

Factory setting0

Calc. factor

Navigation

Expert → Application → Maths → Maths x → Totalization → Calc. factor
Direct access code: 400054-00x
Examples: Maths 1: 400054-000; Maths 4: 400054-003

Description

Factor for calculating the integrated value(e.g. the transmitter delivers l/s -> totalizationbase = second -> engineering unit required is m³ --> enter factor 0.001)
Only visible if totalization = Yes

User entry

Number (max. 8 digits)

Factory setting

1.0

Totalizer

Navigation

Endress+Hauser

Expert → Application → Maths → Maths x → Totalization → Totalizer
Direct access code: 400055-00x
Examples: Maths 1: 400055-000; Maths 4: 400055-003
155

Appendix

Ecograph T, RSG35

Description

Initial setting for the totalizer. Useful when continuing measurements recorded to datewith an (electro)-mechanical counter.
Only visible if totalization = Yes

User entry

Number (max. 15 digits)

Factory setting0"Fault mode" submenu
Navigation
Description

Expert → Application → Maths → Maths x → Fault mode
Contains settings that define how this channel is to react under fault conditions (e.g. if an
input channel has a cable open circuit or there is division by 0).

On error

Navigation

Expert → Application → Maths → Maths x → Fault mode → On error
Direct access code: 400060-00x
Examples: Maths 1: 400060-000; Maths 4: 400060-003

Description

Configure what value the device should continue working with (for calculations) if themeasured value is not valid (e.g. cable open circuit).

Selection

Invalid calculation, Error value

Factory setting

Invalid calculation

Error value

Navigation

Expert → Application → Maths → Maths x → Fault mode → Error value
Direct access code: 400061-00x
Examples: Maths 1: 400061-000; Maths 4: 400061-003

Description

The device continues calculating with this value in the event of an error.
Only visible if on error = Error value

User entry

Number (max. 8 digits)

Factory setting0

Copy settings

156

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Application → Maths → Maths x → Copy settings
Direct access code: 400200-00x
Examples: Maths 1: 400200-000; Maths 4: 400200-003

Description

Copies settings from actual channel to selected channel.

Selection

No, In maths. channel x
Users can choose from all the available maths channels.

Factory settingNo"Signal analysis" submenu
Navigation

Expert → Application → Signal analysis

Description

Contains settings for signal analysis (saving).

Analysis x

Navigation

Description

Expert → Application → Signal analysis → Analysis x
Direct access code: 44000x-000
Examples: Analysis 1: 440000-000; Analysis 4: 440003-000
For the set time frame, determines the minimum, maximum and average value orquantities and operating times.
If the "Controlled externally" option is to be used, a digital input or a maths channelmust be set to "Function = Control input" and "Action = Start/stop analysis x".
Only analysis 1 can be configured; analyses 2-4 are permanently set to daily analysis,monthly analysis and annual analysis

Selection

Switched off, Controlled externally, 1min, 2min, 3min, 4min, 5min, 10min, 15min, 30min,
1h, 2h, 3h, 4h, 6h, 8h, 12h

Factory setting

Switched off

Synchron. time

Navigation

Expert → Application → Signal analysis → Synchron. time
Direct access code: 440004-000

Description

Time for completing the signal analysis.
If, for example, 07:00 is set up, then the daily analysis will run from 07:00 of the actual
day until 07:00 of the following day.

User entry

Time

Factory setting

00:00

Endress+Hauser

157

Appendix

Ecograph T, RSG35

Reset to zero

Navigation

Expert → Application → Signal analysis → Reset to zero
Direct access code: 440007-000

Description

Reset analysis.
Note: Should only be executed after the device has taken over the setup.

Selection

Please select, Analysis x, Totalizer, All

Factory setting

Please selectReset channel

Navigation

Expert → Application → Signal analysis → Reset channel
Direct access code: 440010-000

Description

Reset analysis of a single channel.
Note: Should only be executed after the device has taken over the setup.

Selection

Please select, Universal input x, Digital input x, Maths x, Limit x, Relay x
All active inputs are available for selection.

Factory setting

Please select"Limits" submenu
Navigation
Description

Expert → Application → Limits
Limit values can monitor the measured values. A relay, for example, can be switched if alimit value is violated.

Add limit value

Navigation

Expert → Application → Limits → Add limit value
Direct access code: 450300-000

Description

Adding a new limit value.

Selection

No, Yes

Factory setting

No

158

Endress+Hauser

Ecograph T, RSG35

Appendix

Delete limit value

Navigation

Expert → Application → Limits → Delete limit value
Direct access code: 450301-000

Description

Deleting a limit value from the list.

Selection

No, Limit value x

Factory setting

No"Limit x" submenu
Navigation
Description

Expert → Application → Limits → Limit x
View or change the setup for the selected alarm set point.
x = place holder for selected limit value

Channel/value

Navigation

Expert → Application → Limits → Limit x → Channel/value
Direct access code: 450000-0xx
Examples: Limit 1: 450000-000; Limit 30: 450000-029

Description

Select which input/calculated value the limit value refers to.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Type

Navigation

Expert → Application → Limits → Limit x → Type
Direct access code: 450001-0xx
Examples: Limit 1: 450001-000; Limit 30: 450001-029

Description

Type of limit value (depends on the input variable).

Selection

Switched off, Upper set point, Lower set point, Analysis x

Endress+Hauser

159

Appendix

Ecograph T, RSG35

Description of the individual set point types
Set point type/function

Description

Hysteresis

For every set point, the switch point can be controlled via a hysteresis.
The hysteresis is set as an absolute value (only positive values) in the unit of the respective channel (e.g. upper
set point = 100 m, hysteresis = 1 m: set point on = 100 m, set point off = 99 m)

Upper set point

The limit value is active if the value exceeds the configured value. The limit value is switched off if the limit
value, including hysteresis, is undershot.

Measured value

Setpoint
Hysteresis

Threshold“off”

tSetpoint “on”
Setpoint “off”

A0010187-EN

Lower set point

The limit is active if the value drops below the configured value. The limit value is switched off if the limit value,
including hysteresis, is exceeded.

Measured value

Threshold“off”

Hysteresis

Setpointt
Setpoint “on”
Setpoint “off”

A0010186-EN

Special case: Hysteresis and delay
for one limit value

In the special case in which the hysteresis and limit value delay are activated, a limit value is switched according
to the following principle. If the hysteresis and the limit value delay are activated, the delay becomes active
when a limit value is exceeded and measures the time from which the value is exceeded. If the measured value
falls below the limit value, the delay is reset. This also occurs if the measured value falls below the limit value,
but continues to be higher than the set hysteresis value. When the limit value is exceeded again, the delay time
becomes active again and starts measuring from 0.

Measured
value

Delay

Delay

Setpoint
Hysteresis
Delay timeis resett
Setpoint
activeSetpoint
not active
A0010193-EN

160

Endress+Hauser

Ecograph T, RSG35

Factory setting

Appendix

Switched off

Identifier

Navigation

Expert → Application → Limits → Limit x → Identifier
Direct access code: 450015-0xx
Examples: Limit 1: 450015-000; Limit 30: 450015-029

Description

Name of the limit for identification purposes.

User entry

Text (max. 16 characters)

Factory setting

Limit xSet point

Navigation

Expert → Application → Limits → Limit x → Set point
Direct access code: 450003-0xx
Examples: Limit 1: 450003-000; Limit 30: 450003-029

Description

Limit value in the set process unit, e.g. in °C, m³/h

User entry

Number (max. 10 digits)

Factory setting0

Hysteresis (abs.)

Navigation

Expert → Application → Limits → Limit x → Hysteresis (abs.)
Direct access code: 450004-0xx
Examples: Limit 1: 450004-000; Limit 30: 450004-029

Description

The alarm condition is only canceled when the signal has changed into the normal
operation range by the preset value.

User entry

Number (max. 8 digits)

Factory setting0Time delay

Endress+Hauser

161

Appendix

Ecograph T, RSG35

Navigation

Expert → Application → Limits → Limit x → Time delay
Direct access code: 450005-0xx
Examples: Limit 1: 450005-000; Limit 30: 450005-029

Description

In order to be interpreted as an alarm the signal must exceed or undercut the preset value
by at least the time set up.

User entry

0 to 99999 s

Factory setting

0s

Switches

Navigation

Expert → Application → Limits → Limit x → Switches
Direct access code: 450006-0xx
Examples: Limit 1: 450006-000; Limit 30: 450006-029

Description

Switches the appropriate output in the limit value state.

Selection

Not used, Relay x

Factory setting

Not used

LV messages

Navigation

Expert → Application → Limits → Limit x → LV messages
Direct access code: 450007-0xx
Examples: Limit 1: 450007-000; Limit 30: 450007-029

Description"Do not acknowledge": Alarm condition is signaled by highlighting the tag name in red (nomessage is output).
"Acknowledge": In the event of an alarm, a message is also displayed. This message thenhas to be acknowledged.

Selection

Do not acknowledge, Acknowledge

Factory setting

Do not acknowledge

Save event

Navigation

Description

162

Expert → Application → Limits → Limit x → Save event
Direct access code: 450008-0xx
Examples: Limit 1: 450008-000; Limit 30: 450008-029
Stores a message in the event log on limit value violation.

Endress+Hauser

Ecograph T, RSG35

Appendix

Selection

No, Yes, only "On" message

Factory setting

Yes

Event text LV on

Navigation

Expert → Application → Limits → Limit x → Event text LV on
Direct access code: 450009-0xx
Examples: Limit 1: 450009-000; Limit 30: 450009-029

Description

This text (including date and time) is shown on the display and/or stored in the event log.
Only available if "LV messages" is set to "Acknowledge" or "Save message" is set to "Yes".
If no text is entered, the device generates its own text (e.g. Analog 1 > 100%).

User entry

Text (max. 22 characters)

Event text LV off

Navigation

Expert → Application → Limits → Limit x → Event text LV off
Direct access code: 450010-0xx
Examples: Limit 1: 450010-000; Limit 30: 450010-029

Description

The same as "Event text LV on", but on return from alarm to normal condition.

User entry

Text (max. 22 characters)

Record duration of LV on

Navigation

Expert → Application → Limits → Limit x → Record duration of LV on
Direct access code: 450011-0xx
Examples: Limit 1: 450011-000; Limit 30: 450011-029

Description

The duration of a set point violation can be recorded. The duration is appended to the"Limit value off" event text (format: <hhhh>h<mm>:<ss>).
Power failure times do not affect the duration. If the set point was violated before thepower failure and is still violated after the power failure, the duration continues.

Selection

No, Yes

Factory setting

No

Save cycle

Endress+Hauser

163

Appendix

Ecograph T, RSG35

Navigation

Description

Expert → Application → Limits → Limit x → Save cycle
Direct access code: 450012-0xx
Examples: Limit 1: 450012-000; Limit 30: 450012-029
Normal: Save in normal store cycle.
Alarm cycle: Fast storage during an alarm violation, e.g. every second. Caution: Requires
higher memory capacity!
• The save cycle is set under signal groups .
• In the event of an alarm violation, all the groups are saved in the alarm cycle.

Selection

Normal, Alarm cycle

Factory setting

NormalDraw help line

Navigation

Expert → Application → Limits → Limit x → Draw help line
Direct access code: 450013-0xx
Examples: Limit 1: 450013-000; Limit 30: 450013-029

Description

The user can configure whether this set point should be displayed in the graphic as a helpline (in the color of the channel).
Note: 4 lines can be shown per channel in a single group.

Selection

No, Yes

Factory settingNo

Copy settings

Navigation

Expert → Application → Limits → Limit x → Copy settings
Direct access code: 450200-0xx
Examples: Limit 1: 450200-000; Limit 30: 450200-029

Description

Copies settings from actual channel to selected channel.

Selection

No, In limit x (all the limit values are displayed)

Factory settingNo"Signal groups" submenu
Navigation

164

Expert → Application → Signal groups

Endress+Hauser

Ecograph T, RSG35

Description

Appendix

Group the analog, digital and/or mathematics channels such that you can call up
important information during operation (e.g. temperatures, signals in plant unit 1).
Maximum 8 channels per group!

"Group x" submenu
Navigation

Expert → Application → Signal groups → Group x

Description

x = place holder for selected group
General settings for displaying the measured value and saving data.

Identifier

Navigation

Expert → Application → Signal groups → Group x → Identifier
Direct access code: 460000-0xx
Examples: Group 1: 460000-000; Group 4: 460000-003

Description

Enter a name for these groups.

User entry

Text (max. 20 characters)

Factory settingGroup x

Save cycle

Navigation

Description

Expert → Application → Signal groups → Group x → Save cycle
Direct access code: 460001-0xx
Examples: Group 1: 460001-000; Group 4: 460001-003
Configure the save cycle with which this group should be saved in normal conditions (seealso set point/save cycle).
The save cycle is independent of the measured value display (see Operating
Instructions).

Selection

Off, 1s, 2s, 3s, 4s, 5s, 10s, 15s, 20s, 30s, 1min, 2min, 3min, 4min, 5min, 10min, 15min,
30min, 1h

Factory setting

1min

Alarm cycle

Endress+Hauser

165

Appendix

Ecograph T, RSG35

Navigation

Expert → Application → Signal groups → Group x → Alarm cycle
Direct access code: 460002-0xx
Examples: Group 1: 460002-000; Group 4: 460002-003

Description

Configure the save cycle with which this group should be saved in an alarm condition (limit
value violation).
Caution: Requires higher memory capacity.

Selection

Off, 1s, 2s, 3s, 4s, 5s, 10s, 15s, 20s, 30s, 1min, 2min, 3min, 4min, 5min, 10min, 15min,
30min, 1h

Factory setting

1min

Display blue

Navigation

Expert → Application → Signal groups → Group x → Display blue
Direct access code: 460003-00x
Examples: Group 1: 460003-000; Group 4: 460003-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460004-00x
Examples: Group 1: 460004-000; Group 4: 460004-003

Description

Please select which data from the selected channel should be displayed.
If the "Everything" option is selected, the device switches cyclically between thevarious values of the channel (instantaneous value, analysis 1 etc.)

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Display black

Navigation

Description
166

Expert → Application → Signal groups → Group x → Display black
Direct access code: 460005-00x
Examples: Group 1: 460005-000; Group 4: 460005-003
Choose which input/calculated variable should be displayed in this group.
Endress+Hauser

Ecograph T, RSG35

Appendix

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460006-0xx
Examples: Group 1: 460006-000; Group 4: 460006-003

Description

Please select which data from the selected channel should be displayed.

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Display red

Navigation

Expert → Application → Signal groups → Group x → Display red
Direct access code: 460007-00x
Examples: Group 1: 460007-000; Group 4: 460007-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460008-0xx
Examples: Group 1: 460008-000; Group 4: 460008-003

Description

Please select which data from the selected channel should be displayed.

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Display green

Endress+Hauser

167

Appendix

Ecograph T, RSG35

Navigation

Expert → Application → Signal groups → Group x → Display green
Direct access code: 460009-00x
Examples: Group 1: 460009-000; Group 4: 460009-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460010-0xx
Examples: Group 1: 460010-000; Group 4: 460010-003

Description

Please select which data from the selected channel should be displayed.

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Display violet

Navigation

Expert → Application → Signal groups → Group x → Display violet
Direct access code: 460011-00x
Examples: Group 1: 460011-000; Group 4: 460011-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460012-0xx
Examples: Group 1: 460012-000; Group 4: 460012-003

Description

Please select which data from the selected channel should be displayed.

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

168

Endress+Hauser

Ecograph T, RSG35

Appendix

Display orange

Navigation

Expert → Application → Signal groups → Group x → Display orange
Direct access code: 460013-00x
Examples: Group 1: 460013-000; Group 4: 460013-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460014-0xx
Examples: Group 1: 460014-000; Group 4: 460014-003

Description

Please select which data from the selected channel should be displayed.

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Display cyan

Navigation

Expert → Application → Signal groups → Group x → Display cyan
Direct access code: 460015-00x
Examples: Group 1: 460015-000; Group 4: 460015-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460016-0xx
Examples: Group 1: 460016-000; Group 4: 460016-003

Description

Please select which data from the selected channel should be displayed.

Endress+Hauser

169

Appendix

Ecograph T, RSG35

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Display brown

Navigation

Expert → Application → Signal groups → Group x → Display brown
Direct access code: 460017-00x
Examples: Group 1: 460017-000; Group 4: 460017-003

Description

Choose which input/calculated variable should be displayed in this group.

Selection

Switched off, Universal input x, Digital input x, Maths x

Factory setting

Switched off

Display

Navigation

Expert → Application → Signal groups → Group x → Display
Direct access code: 460018-0xx
Examples: Group 1: 460018-000; Group 4: 460018-003

Description

Please select which data from the selected channel should be displayed.

Selection

Instantaneous value/state, Analysis x, Totalizer, Everything

Factory setting

Instantaneous value/state

Grid divisions

Navigation

Expert → Application → Signal groups → Group x → Grid divisions
Direct access code: 460019-0xx
Examples: Group 1: 460019-000; Group 4: 460019-003

Description

Indicates the number of lines ("amplitude grid") that should be displayed. Example: displayof 0 to 100%: select 10 divisions, display 0 to 14pH: select 14 divisions.

Selection

1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20

Factory setting

10

Zoom

170

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Application → Signal groups → Group x → Zoom
Direct access code: 460028-0xx
Examples: Group 1: 460028-000; Group 4: 460028-003

Description

Defines the zoom that is shown in "Curves" or "Waterfall" display mode. This setting does
not affect other display modes (e.g. Curves in range, Bar graph etc.).

Selection

Do not display, Scroll display, Display blue, Display black, Display red, Display green,
Display violet, Display orange, Display cyan, Display brown

Factory setting

Do not display"E-mail" submenu
Navigation

Expert → Application → E-mailIn the case of the "Telealarm" option under
Expert → Application → Telealarm → General → Setup e-mailDescription

Contains settings required if alarms are to be transmitted by e-mail.
Test the e-mail settings under Diagnostics → Simulation → E-mail.

SMTP hostNavigation

Expert → Application → E-mail → SMTP host
Direct access code: 510062-000

Description

Enter your SMTP host here. If necessary, contact your network administrator or e-mailprovider.

User entry

Text (max. 40 characters)

Server requires SSL

Navigation

Expert → Application → E-mail → Server requires SSL
Direct access code: 510061-000

Description

Specify whether the e-mail server requires a secure connection (SSL).
STARTTLS: Runs on the same TCP port as unencrypted SMTP (port 25 or 587).
SMTPS: Completely encrypted with own TCP port (465).
If necessary, contact your network administrator or e-mail provider.

Selection

No, Yes (SMTPS), Yes (STARTTLS)

Factory setting

No

Endress+Hauser

171

Appendix

Ecograph T, RSG35

Port

Navigation

Expert → Application → E-mail → Port
Direct access code: 510063-000

Description

Enter your SMTP port here. If necessary, contact your network administrator or e-mailprovider.

User entry

Number (max. 4 digits)

Factory setting

25

SenderNavigation

Description

Expert → Application → E-mail → Sender
Direct access code: 510064-000
Enter the e-mail address of the device here (this text appears as the sender of the e-mail).
If necessary, contact your network administrator or e-mail provider.
If a valid e-mail address is not configured this might cause e-mail transmission
problems, depending on the particular provider.

User entry

Text (max. 60 characters)

User name

Navigation

Expert → Application → E-mail → User name
Direct access code: 510066-000

Description

Configure the user name of the e-mail account here. If necessary, contact your network
administrator or e-mail provider.

User entry

Text (max. 60 characters)

Password

Navigation

Expert → Application → E-mail → Password
Direct access code: 510067-000

Description

Enter the password for authentication here. If necessary, contact your network
administrator or e-mail provider.

User entry

Text (max. 22 characters)

172

Endress+Hauser

Ecograph T, RSG35

Appendix"E-mail addresses" submenu
Navigation
Description

Expert → Application → E-mail → E-mail addresses
Here, enter all the e-mail addresses messages should be sent to in the event of an alarm.
Assignment to the alarms is performed later on.

E-mail address x

Navigation

Description

Expert → Application → E-mail → E-mail addresses → E-mail address x
Direct access code:
E-mail address 1: 510080-000...
E-mail address 5: 510084-000
Here, enter an e-mail address a message should be sent to.
Assignment to the alarms is performed later on.

User entry

Text (max. 60 characters)"Limit value violations" submenu
Navigation
Description

Expert → Application → E-mail → Limit value violations
Specify who should receive e-mails when limit value violations occur (both on and off
messages).
Only for limit values where "Save message" is set to "Yes".

Recipient x

Navigation

Expert → Application → E-mail → Limit value violations → Recipient x
Direct access code:
Recipient 1: 510110-000; Recipient 2: 510111-000

Description

Select who should receive the e-mail.

Selection

Not used, E-mail address x

Factory setting

Not used"On/off messages" submenu

Endress+Hauser

173

Appendix

Ecograph T, RSG35

Navigation
Description

Expert → Application → E-mail → On/off messages
Specify who should receive e-mails when "on"/"off" messages occur (for digital inputs ormaths channels).
Only for inputs where "Save event" is set to "Yes".

Recipient x

Navigation

Expert → Application → E-mail → On/off messages → Recipient x
Direct access code:
Recipient 1: 510115-000; Recipient 2: 510116-000

Description

Select who should receive the e-mail.

Selection

Not used, E-mail address x

Factory setting

Not used"Errors (Fxxx/Sxxx)" submenu
Navigation
Description

Expert → Application → E-mail → Errors (Fxxx/Sxxx)
Specify who should receive e-mails when errors occur (Fxxx and Sxxx messages).

Recipient x

Navigation

Expert → Application → E-mail → Errors → Recipient x
Direct access code:
Recipient 1: 510120-000; Recipient 2: 510121-000

Description

Select who should receive the e-mail.

Selection

Not used, E-mail address x

Factory setting

Not used"Maintenance required" (submenu)
Navigation
Description

Expert → Application → E-mail → Maintenance required
Specify who should receive e-mails when maintenance is required (Mxxx messages).

Recipient x

174

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Application → E-mail → Maintenance required → Recipient x
Direct access code:
Recipient 1: 510130-000; Recipient 2: 510131-000

Description

Select who should receive the e-mail.

Selection

Not used, E-mail address x

Factory setting

Not used"WebDAV Client" submenu

Navigation
Description

Expert → Application → WebDAV Client
All recorded data are transmitted to an external WebDAV server (e.g. NAS). The format
can be specified or selected via "Setup → Advanced setup → System → External memory-> Save as".
Parameter

Description

Direct access code

EnableSwitch the WebDAV client functionality on/off. When active, the device
copies the saved measured values automatically to the configured server.

472000-000 Only possible using the Ethernet interface!
Options: No, Yes, Yes (SSL)
Factory setting: No
IP addressEnter the IP address of the WebDAV server here.



472001-000

A DNS name can also be used.

User entry: IP address
Factory setting: 0.0.0.0
PortThis communication port is used to communicate with the WebDAV
Server.

472002-000

your network is protected by a firewall, this port may have to be Ifenabled.
Please contact your network administrator if this is thecase.
User entry: Numbers (max. 5 digits)
Factory setting: 80
User name

Input of the user name that can access the WebDAV server.
User entry: Text (max. 20 characters)

472004-000

Password

Password for accessing the WebDAV server.
User entry: Text (max. 20 characters)

472007-000

Directory

Enter the directory in which the data should be saved.
User entry: Text (max. 120 characters)

472005-000

Save as"Protected format": All data is stored in a manipulation protectedencrypted format. This data can only be visualized by the PC analysis
software supplied.
"Open format": Data is stored in a CSV format, this can be opened by anumber of different programs (for e.g. MS Excel) (Attention: no
manipulation security).
Options: Protected format, Open format (*.csv)
Factory setting: Protected format

472010-000

Test the WebDAV client settings under "Diagnostics → Simulation → WebDAV
client".
Endress+Hauser

175

Appendix

Ecograph T, RSG35

16.1.6"Diagnostics" submenu

Device information and service functions for a quick device check.
Only some of the diagnostic functions are available under Expert → Diagnostics! Forother functions, see the Main menu → Diagnostics

Actual diagnostics

Navigation

Description

Expert → Diagnostics → Actual diagnostics
Direct access code: 050000-000
Displays the current diagnosis message.

Last diagnostics

Navigation

Description

Expert → Diagnostics → Last diagnostics
Direct access code: 050005-000
Displays the last diagnosis message.

Last restart

Navigation

Description

Expert → Diagnostics → Last restart
Direct access code: 050010-000
Information as to when the device was last restarted (e.g. due to a power failure).

"Event log" submenu
Navigation
Description

Expert → Diagnostics → Event log
Events such as alarm set point infringement and power failure are listed in the correcttime sequence.

"Device information" submenu
Navigation
Description

Expert → Diagnostics → Device information
Displays important device information.

Device tag

176

Endress+Hauser

Ecograph T, RSG35

Appendix

Navigation

Expert → Diagnostics → Device information → Device tag
Direct access code: 000031-000

Description

Individual device tag name/unit identifier (max. 32 characters).

Serial number

Navigation

Description

Expert → Diagnostics → Device information → Serial number
Direct access code: 000027-000
Individual serial number of the device. Please provide these details when ordering spareparts or asking any questions about the unit.

Order code

Navigation

Description

Expert → Diagnostics → Device information → Order code
Direct access code: 000029-000
Displays the order code.
The order code indicates the attribute of all the features of the product structure for thedevice and thus uniquely identifies the device. It can also be found on the nameplate.
Uses of the order code
• To order an identical replacement device.
• To check the ordered device features using the delivery note.

Firmware version

Navigation

Description

Expert → Diagnostics → Device information → Firmware version
Direct access code: 000026-000
Displays the installed firmware version of the device. Please send these details with anyquestions about the unit.

ENP version

Navigation

Description

Endress+Hauser

Expert → Diagnostics → Device information → ENP version
Direct access code: 000032-000
Displays the version of the electronic nameplate. Please send these details with anyquestions about the unit.

177

Appendix

Ecograph T, RSG35

ENP device name

Navigation

Description

Expert → Diagnostics → Device information → ENP device name
Direct access code: 000020-000
Displays the ENP device name (electronic name plate). Please send these details with anyquestions about the unit.

Device name

Navigation

Description

Expert → Diagnostics → Device information → Device name
Direct access code: 000021-000
Displays the device name. Please send these details with any questions about the unit.

Manufacturer ID

Navigation

Description

Expert → Diagnostics → Device information → Manufacturer ID
Direct access code: 000022-000
Displays the manufacturer ID. Please send these details with any questions about the unit.

Manufacturer name

Navigation

Description

Expert → Diagnostics → Device information → Manufacturer name
Direct access code: 000023-000
Displays the manufacturer name. Please send these details with any questions about theunit.

Firmware

Navigation

Description

178

Expert → Diagnostics → Device information → Firmware
Direct access code: 009998-000
Displays the installed firmware of the device. Please send these details with any questions
about the unit.

Endress+Hauser

Ecograph T, RSG35

Appendix"Simulation" submenu
Navigation
Description

Expert → Diagnostics → Simulation
Settings for simulation mode.

Operating mode

Navigation

Expert → Diagnostics → Simulation → Operating mode
Direct access code: 010010-000

Description

Normal operation: Unit plots the signals from the connected measurement points.
Simulation: Instead of operating with the real measurement points the signals are
simulated (using the actual settings).

Selection

Normal operation, Simulation

Factory setting

Normal operation

Endress+Hauser

179

IndexEcograph T, RSG35

Index
0…9

1 hour= (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 125
1 second= (parameter) . . . . . . . . . . . . . . . . . . . . . . 125

A

Access code (parameter) . . . . . . . . . . . . . . . . . . . . . . 98
Acknowledging messages (parameter) . . . . . . . . . . . 103
Action (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 125
Activation code (parameter) . . . . . . . . . . . . . . . . . . 105
Actual diagnostics (parameter) . . . . . . . . . . . . . . . . 176
Actual value (parameter) . . . . . . . . . . . . . . . . . 116, 117
Add input (parameter) . . . . . . . . . . . . . . . . . . . 107, 122
Add limit value (parameter) . . . . . . . . . . . . . . . . . . . 158
Admin ID (parameter) . . . . . . . . . . . . . . . . . . . . . . . 137
Admin password (parameter) . . . . . . . . . . . . . . . . . 138
Administrator (parameter) . . . . . . . . . . . . . . . . . . . 137
Administrator, ID, password (parameter) . . . . . . . . . 100
Alarm cycle (parameter) . . . . . . . . . . . . . . . . . . . . . 165
Alarm response (parameter) . . . . . . . . . . . . . . . . . . 104
Analysis x (parameter) . . . . . . . . . . . . . . . . . . . . . . 157
Application (parameter) . . . . . . . . . . . . . . . . . . . . . 107
Application (submenu) . . . . . . . . . . . . . . . . . . . . . . 145
Authentication (submenu) . . . . . . . . . . . . . . . . . . . . . 99

B

Baud rate (parameter) . . . . . . . . . . . . . . . . . . . . . . . 142
Begin summer time (parameter) . . . . . . . . . . . . . . . . 94

C

Cable open circuit (parameter) . . . . . . . . . . . . . . . . . 119
Calc. factor (parameter) . . . . . . . . . . . . . . . . . . 118, 155
CE mark . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
CE mark (declaration of conformity) . . . . . . . . . . . . . . . 9
Change date/time (submenu) . . . . . . . . . . . . . . . . . . 92
Channel ident. (parameter) . . . . . . . . . . . . . . . 109, 123
Channel/value (parameter) . . . . . . . . . . . . . . . . . . . 159
Clear memory (parameter) . . . . . . . . . . . . . . . . . . . . 91
Communication
Ethernet TCP/IP . . . . . . . . . . . . . . . . . . . . . . . . . . 19
Communication (parameter) . . . . . . . . . . . . . . . . . . 106
Communication (submenu) . . . . . . . . . . . . . . . . . . . 130
Comparison point (parameter) . . . . . . . . . . . . . . . . . 114
Comparison temp. (parameter) . . . . . . . . . . . . . . . . 114
Configuration softwareFieldCare . . . . . . . . . . . . . . . . 38
Configuration Web server (submenu) . . . . . . . . . . . . 134
Connection (parameter) . . . . . . . . . . . . . . . . . . . . . 109
Connection quality (parameter) . . . . . . . . . . . . . . . . 139
Copy settings (parameter) . . . . . . . . 122, 129, 156, 164
Correction RPT (parameter) . . . . . . . . . . . . . . . . . . . 115
CSV settings (parameter) . . . . . . . . . . . . . . . . . . . . . 102
Current date/time (parameter) . . . . . . . . . . . . . . . . . 92

D

Damping (parameter) . . . . . . . . . . . . . . . . . . . . . . . 113
Date (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 95, 96
Date format (parameter) . . . . . . . . . . . . . . . . . . . . . . 91
Date/time (parameter) . . . . . . . . . . . . . . . . . . . 93, 102
180

Date/time (submenu) . . . . . . . . . . . . . . . . . . . . . . . . 92
Date/time setup (submenu) . . . . . . . . . . . . . . . . . . . . 91
Day (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 94, 96
Decimal point (parameter) . . . . . . . . . . . . . . . . 111, 124
Decimal separator (parameter) . . . . . . . . . . . . . . . . . 89
Declaration of Conformity . . . . . . . . . . . . . . . . . . . . . . 9
Delete input (parameter) . . . . . . . . . . . . . . . . . 107, 122
Delete limit value (parameter) . . . . . . . . . . . . . . . . . 159
Description 'H' (parameter) . . . . . . . . . . . . . . . . 126, 148
Description 'L' (parameter) . . . . . . . . . . . . . . . . 127, 149
Device address (parameter) . . . . . . . . . . . . . . . . . . . 143
Device info (submenu) . . . . . . . . . . . . . . . . . . . . . . . 176
Device name . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Device options (submenu) . . . . . . . . . . . . . . . . . . . . 105
Device tag . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 176
Device tag (parameter) . . . . . . . . . . . . . . . . . . . . . . . 88
DHCP (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 132
Diagnostic messages . . . . . . . . . . . . . . . . . . . . . . . . . 59
Diagnostics (submenu) . . . . . . . . . . . . . . . . . . . . . . 176
Digital inputs (submenu) . . . . . . . . . . . . . . . . . . . . . 122
Direct access (parameter) . . . . . . . . . . . . . . . . . . . . . 88
Disable port (parameter) . . . . . . . . . . . . . . . . . . . . . 133
Display (parameter) . . . . . . . . . 166, 167, 168, 169, 170
Display black (parameter) . . . . . . . . . . . . . . . . . . . . 166
Display blue (parameter) . . . . . . . . . . . . . . . . . . . . . 166
Display brown (parameter) . . . . . . . . . . . . . . . . . . . 170
Display cyan (parameter) . . . . . . . . . . . . . . . . . . . . . 169
Display green (parameter) . . . . . . . . . . . . . . . . . . . . 167
Display orange (parameter) . . . . . . . . . . . . . . . . . . . 169
Display red (parameter) . . . . . . . . . . . . . . . . . . . . . . 167
Display violet (parameter) . . . . . . . . . . . . . . . . . . . . 168
Domain Name System (DNS) (parameter) . . . . . . . . 133
Draw help line (parameter) . . . . . . . . . . . . . . . . . . . 164

EE-mail (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 171
E-mail address x (parameter) . . . . . . . . . . . . . . . . . . 173
E-mail addresses (submenu) . . . . . . . . . . . . . . . . . . 173
End summer time (parameter) . . . . . . . . . . . . . . . . . . 95
Engineering unit (parameter) . . . . . . . . . . . . . 110, 124
ENP device name . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
ENP version . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 177
Error (Fxxx/Sxxx) (submenu) . . . . . . . . . . . . . . . . . . 174
Error messages . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Error value (parameter) . . . . . . . . . . . . . . . . . . 121, 156
Ethernet . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Ethernet configuration (submenu) . . . . . . . . . . . . . . 131
Event log (submenu) . . . . . . . . . . . . . . . . . . . . . . . . 176
Event message (parameter) . . . . . . . . . . . . . . . 127, 149
Event text H->L (parameter) . . . . . . . . . . . . . . 128, 150
Event text L->H (parameter) . . . . . . . . . . . . . . 128, 150
Event text LV off (parameter) . . . . . . . . . . . . . . . . . 163
Event text LV on (parameter) . . . . . . . . . . . . . . . . . . 163
Expert (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 88
External memory (submenu) . . . . . . . . . . . . . . . . . . 100

Endress+Hauser

Ecograph T, RSG35

F

Fault mode (submenu) . . . . . . . . . . . . . . . . . . . 119, 156
Fault switching (parameter) . . . . . . . . . . . . . . . . . . . 89
Field Data Manager (FDM) analysis software
Function scope . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Fieldbus (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 106
FieldCare/DeviceCare configuration software
Function scope . . . . . . . . . . . . . . . . . . . . . . . . . . . 31
Firmware (parameter) . . . . . . . . . . . . . . . . . . . . . . . 178
Firmware update (parameter) . . . . . . . . . . . . . . . . . 135
Firmware version (parameter) . . . . . . . . . . . . . . . . . 177
Formula (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 146
Formula editor (parameter) . . . . . . . . . . . . . . . . . . . 151
Formula editor (submenu) . . . . . . . . . . . . . . . . . . . . 151
Function (parameter) . . . . . . . . . . . . . . . . . . . . . . . 123
Function USB-B (parameter) . . . . . . . . . . . . . . . . . . 131

G

Gateway (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 133
Get timeout (parameter) . . . . . . . . . . . . . . . . . . . . . 139
Grid divisions (parameter) . . . . . . . . . . . . . . . . . . . . 170
Group x (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 165

H

Hysteresis (abs.) (parameter) . . . . . . . . . . . . . . . . . . 161

I

Identifier (parameter) . . . . . . . . . . . . . . . 130, 161, 165
Input factor in (parameter) . . . . . . . . . . . . . . . . . . . 124
Inputs (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 107
IP address (parameter) . . . . . . . . . . . . . . . . . . . . . . 132

K

Keyboard layout (parameter) . . . . . . . . . . . . . . . . . . . 89

L

Language (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 88
Last diagnostics (parameter) . . . . . . . . . . . . . . . . . . 176
Last restart (parameter) . . . . . . . . . . . . . . . . . . . . . 176
Limit value violations (submenu) . . . . . . . . . . . . . . . 173
Limit x (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 159
Limits (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 158
Lock hardware (parameter) . . . . . . . . . . . . . . . . . . . . 99
Lock operation (parameter) . . . . . . . . . . . . . . . . . . . . 90
Low flow cut off (parameter) . . . . . . . . . . . . . . 118, 155
Lower error value (parameter) . . . . . . . . . . . . . . . . . 120
Lower frequency (parameter) . . . . . . . . . . . . . . . . . . 111
LV messages (parameter) . . . . . . . . . . . . . . . . . . . . 162

M

MAC address (parameter) . . . . . . . . . . . . . . . . . . . . 132
Maintenance required (submenu) . . . . . . . . . . . . . . 174
Manufacturer ID (parameter) . . . . . . . . . . . . . . . . . . 178
Manufacturer name . . . . . . . . . . . . . . . . . . . . . . . . 178
Maths (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Maths x action (parameter) . . . . . . . . . . . . . . . . . . . 147
Maths x channel ident. (parameter) . . . . . . . . . . . . . 145
Maths x decimal point (parameter) . . . . . . . . . . . . . 147
Maths x engineering unit (parameter) . . . . . . . . . . . 147
Maths x function (parameter) . . . . . . . . . . . . . . . . . 145
Endress+Hauser

Index

Maths x plot type (parameter) . . . . . . . . . . . . . . . . . 147
Meas. range end (parameter) . . . . . . . . . . . . . . 112, 116
Meas. val. without login (parameter) . . . . . . . . . . . . 136
Meas.val. corrct. (submenu) . . . . . . . . . . . . . . . . . . . 115
Memory build-up (parameter) . . . . . . . . . . . . . . . . . 101
Messages (submenu) . . . . . . . . . . . . . . . . . . . . . . . . 103
Modbus (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 143
Modbus RTU/(TCP/IP) . . . . . . . . . . . . . . . . . . . . . . . . 34
Modbus Slave (submenu) . . . . . . . . . . . . . . . . . . . . . 143
Modbus Slave baud rate (parameter) . . . . . . . . . . . . 144
Modbus Slave parity (parameter) . . . . . . . . . . . . . . . 144
Modbus Slave serial interface (submenu) . . . . . . . . . 144
Modbus Slave stop bits (parameter) . . . . . . . . . . . . . 145
Month (parameter) . . . . . . . . . . . . . . . . . . . . . . . 94, 96

N

NAMUR NE 43 (parameter) . . . . . . . . . . . . . . . . . . . 119
NT/ST changeover (parameter) . . . . . . . . . . . . . . . . . 93
NT/ST changeover (submenu) . . . . . . . . . . . . . . . . . . 93
NT/ST region (parameter) . . . . . . . . . . . . . . . . . . . . . 93

O

Occurrence (parameter) . . . . . . . . . . . . . . . . . . . . 94, 95
OFF daily from (parameter) . . . . . . . . . . . . . . . . . . . 104
Offset (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 115
ON daily from (parameter) . . . . . . . . . . . . . . . . . . . 104
On error (parameter) . . . . . . . . . . . . . . . . . . . . 121, 156
On/off messages (submenu) . . . . . . . . . . . . . . . . . . 173
OPC port (parameter) . . . . . . . . . . . . . . . . . . . . . . . 134
OPC server
Function scope . . . . . . . . . . . . . . . . . . . . . . . . . . . 31
Operating mode (parameter) . . . . . . . . . . . . . . 130, 179
Operation options
Local operation . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Operating tool . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Overview . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Operational time (parameter) . . . . . . . . . . . . . . . . . 102
Operator (parameter) . . . . . . . . . . . . . . . . . . . . . . . 137
Operator ID (parameter) . . . . . . . . . . . . . . . . . . . . . 137
Operator password (parameter) . . . . . . . . . . . . . . . . 137
Operator, ID, password (parameter) . . . . . . . . . . . . . . 99
Order code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 177
Other standards and guidelines . . . . . . . . . . . . . . . . . 86
Output (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Overview of symbols . . . . . . . . . . . . . . . . . . . . . . . . . 28

P

Paper size (parameter) . . . . . . . . . . . . . . . . . . . . . . . 90
Parity (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 142
Password (parameter) . . . . . . . . . . . . . . . . . . . . . . . 172
Ping interval (parameter) . . . . . . . . . . . . . . . . . . . . 140
Ping retry (parameter) . . . . . . . . . . . . . . . . . . . . . . . 141
Ping timeout (parameter) . . . . . . . . . . . . . . . . . . . . 140
Plot type (parameter) . . . . . . . . . . . . . . . . . . . . . . . 109
Poll timeout (parameter) . . . . . . . . . . . . . . . . . . . . . 141
Port (parameter) . . . . . . . . . . . . . . . 134, 135, 144, 172
PRESET (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 90
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
181

IndexProtected by (parameter) . . . . . . . . . . . . . . . . . . . . . . 98
Protocol (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 142
Pulse counter (parameter) . . . . . . . . . . . . . . . . . . . . 110
Pulse value (parameter) . . . . . . . . . . . . . . . . . . 111, 124
Put timeout (parameter) . . . . . . . . . . . . . . . . . . . . . 140

R

Range (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 108
Range start (parameter) . . . . . . . . . . . . . . . . . 111, 115
Recipient x (parameter) . . . . . . . . . . . . . . . . . . 173, 174
Record duration (parameter) . . . . . . . . . . . . . . 128, 150
Record duration of LV on (parameter) . . . . . . . . . . . 163
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . . . 8
Relay (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Remote control (parameter) . . . . . . . . . . . . . . . . . . . 136
Reset channel (parameter) . . . . . . . . . . . . . . . . . . . . 158
Reset to zero (parameter) . . . . . . . . . . . . . . . . . . . . 158
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 69

S

Save as (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 100
Save cycle (parameter) . . . . . . . . . . . . . . . . . . . 163, 165
Save event (parameter) . . . . . . . . . . 121, 127, 149, 162
Screensaver (parameter) . . . . . . . . . . . . . . . . . . . . . 103
Screensaver (submenu) . . . . . . . . . . . . . . . . . . . . . . 103
SD card (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 101
Security (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 98
Sender (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 172
Separator for CSV (parameter) . . . . . . . . . . . . . . . . . 102
Serial interface (submenu) . . . . . . . . . . . . . . . . . . . . 141
Serial number . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 177
Server requires SSL (parameter) . . . . . . . . . . . . . . . . 171
Service (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 138
Service ID (parameter) . . . . . . . . . . . . . . . . . . . . . . . 138
Service password (parameter) . . . . . . . . . . . . . . . . . 138
Service, ID, password (parameter) . . . . . . . . . . . . . . 100
Set point (parameter) . . . . . . . . . . . . . . . . . . . . . . . 161
Set point code (parameter) . . . . . . . . . . . . . . . . . . . . 98
Set timeout (parameter) . . . . . . . . . . . . . . . . . . . . . 139
Setup (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Setup via Web server . . . . . . . . . . . . . . . . . . . . . . . . . 37
Signal (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 108
Signal analysis (submenu) . . . . . . . . . . . . . . . . . . . . 157
Signal groups (submenu) . . . . . . . . . . . . . . . . . . . . . 164
Simulation (submenu) . . . . . . . . . . . . . . . . . . . . . . . 179
Slot 1 (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 105
Slot 2 (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Slot 3 (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 106
SMTP host (parameter) . . . . . . . . . . . . . . . . . . . . . . 171
SNTP (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
SNTP (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
SNTP server 1 (parameter) . . . . . . . . . . . . . . . . . . . . 97
SNTP server 2 (parameter) . . . . . . . . . . . . . . . . . . . . 97
Staff
Requirements . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Stop bits (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 142
Structure of the operating menu . . . . . . . . . . . . . . 24, 25
Subnetmask (parameter) . . . . . . . . . . . . . . . . . . . . . 132
Swap mouse buttons (parameter) . . . . . . . . . . . . . . . . 90
182

Ecograph T, RSG35

Switches (parameter) . . . . . . . . . . . . . . . . . . . 131, 162
Switches relay (parameter) . . . . . . . . 101, 103, 126, 148
Symbols
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Operating menus . . . . . . . . . . . . . . . . . . . . . . . . . 29
Synchron. time (parameter) . . . . . . . . . . . . . . . . . . . 157
System (submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 88

T

Target value (parameter) . . . . . . . . . . . . . . . . . . . . . 116
Temp. unit (parameter) . . . . . . . . . . . . . . . . . . . . . . . 89
Text entry . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
The result is (parameter) . . . . . . . . . . . . . . . . . . . . . 146
Time (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 95, 96
Time base (parameter) . . . . . . . . . . . . . . . . . . . . . . 109
Time delay (parameter) . . . . . . . . . . . . . . 120, 125, 161
Time format (parameter) . . . . . . . . . . . . . . . . . . . . . . 92
Timeout (parameter) . . . . . . . . . . . . . . . . . . . . . . . . 130
Timeout fieldbus (parameter) . . . . . . . . . . . . . . . . . 131
Timeouts (submenu) . . . . . . . . . . . . . . . . . . . . . . . . 139
Totalization (parameter) . . . . . . . . . . . . . . . . . 117, 154
Totalization (submenu) . . . . . . . . . . . . . . . . . . 117, 154
Totalization base (parameter) . . . . . . . . . . . . . 117, 154
Totalizer (parameter) . . . . . . . . 114, 119, 129, 151, 155
Troubleshooting
Alarm relay . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Type (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 159
Type RS232/RS485 (parameter) . . . . . . . . . . . . . . . 141

U

Unit (parameter) . . . . . . . . . . . . . . . . . . . . . . . 118, 154
Unit/dimension counter (parameter) . . . . . . . . . . . . 110
Universal input x (submenu) . . . . . . . . . . . . . . . . . . 108
Universal inputs (submenu) . . . . . . . . . . . . . . . . . . . 107
Upper error value (parameter) . . . . . . . . . . . . . . . . . 120
Upper frequency (parameter) . . . . . . . . . . . . . . . . . . 112
User name (parameter) . . . . . . . . . . . . . . . . . . . . . . 172
UTC time zone (parameter) . . . . . . . . . . . . . . . . . . . . 92

W

Warning at (parameter) . . . . . . . . . . . . . . . . . . . . . 101
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43
Function scope . . . . . . . . . . . . . . . . . . . . . . . . . . . 31
Web server (parameter) . . . . . . . . . . . . . . . . . . . . . 134
Web server authentication (submenu) . . . . . . . . . . . 136
WebDAV Client (submenu) . . . . . . . . . . . . . . . . . . . 175
WebDAV server (parameter) . . . . . . . . . . . . . . . . . . 136
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

Z

Zoom (parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 170
Zoom end (parameter) . . . . . . . . . . . . . . . . . . . 113, 151
Zoom start (parameter) . . . . . . . . . . . . . . . . . . 113, 150

Endress+Hauser

*71543128*
71543128www.addresses.endress.com