Digital force gauges
RZE series catalog

Digital force gauges vol.1.04

Digital force gauges
RZE series

The RZE series is a new customer-friendly model that follows
the functionality of conventional digital push-pull gauges but
greatly improves operability.
They come with such convenient functions as data output
by use of a newly adopted USB and charging and they also
feature quick and easy operations such as changing sampling
rates, 3 units, printer output and display reversing.
This digital force gauge goes beyond conventional thinking,with a shape that fits your hand and operation buttons.

Standard Accessories

015B

014B

013B

012B

011B

016B

Standard Specifications
Measurement unit

N, Kgf, lb selectable

Accuracy

Within ±0.2% of rated capacity

Allowable overload

120% of rated capacity (Overload alarm at about 110%)

Measurement system

Track mode/Peak hold mode/
Compression-tension peak hold mode selectable

Indication update cycle

1, 2, 5, 10, 20 times/sec. selectable

Sampling cycle

1ms, 5ms, 16ms, 50ms, 125ms, 250ms selectable

Communication output

USBAnalog output±2V/F.S.

Printer output

SANEI thermal printer, MITSUTOYO DIGIMATIC output

Operating temperature range

0 to +40ºC

Guaranteed temperature range+5 to +40ºC

Continuous operation time

30 hours after full charge

Power supply

Dedicated nickel-metal hydride battery
Dedicated AC adaptor: MODEL-780 (100 to 240VAC)
Dedicated USB cable: MODEL-RZ-USB

Dimensions

W68 x D40 x H232mm

WeightApprox. 375g

Model List
MODEL

RZE-1

RZE-2

RZE-5

RZE-10

RZE-20RZE-50

RZE-100

Capacity

10N(1kgf)

20N(2kgf)

50N(5kgf)

100N(10kgf)200N(20kgf)

500N(50kgf)

1000N(100kgf)

Indication range

0.001-10.000N(0.1gf-1.0000kgf)

0.01-20.00N(1gf-2.000kgf)

0.01-50.00N(1gf-5.000kgf)

0.01-100.00N(1gf-10.000kgf)

0.1-200.0N(10gf-20.00kgf)

0.1-500.0N(10gf-50.00kgf)

0.1-1000.0N(10gf-100.00kgf)

Minimum display unit

0.001N (0.1gf)

0.01N (1gf)

Optional cables (Dedicated for RZE series)
RZE-OP-1

I/O cable

RZE-OP-2

I/O cable

RZE-OP-3

Cable for printer DP-1VR MITSUTOYO madeRZE-OP-4

Cable for line thermal printer Sanei Electric

0.1N (10gf)

If you only want stand control, please purchase RZE-OP-1.
If you wish to have comparator output or analog output in
addition to the above functions, please purchase RZE-OP-2.

RZ-3000 (Dedicated software for RZE series)
This is dedicated software designed to store data acquiredby the RZE Series on a PC. It acquires and sends real-timedata to a PC to show waveform graphs and maximum values.
Measurement data can be output in CSV files and output CSV
files can be opened using Excel. The standard interface is theUSB interface. (The USB cable is included with the RZE force
gauge as a standard accessory.)

Software for Excel

Real-time waveform

Recommended Environment
Microsoft Windows 7,8,10,11 Japanese or English version
Memory: 2GB or more
CD-ROM drive x 1 (when installing this software)

Vertical & Horizontal manual Stand

Vertical & Horizontal motorized Stand

MODEL-2256

MODEL-2257

Grip mounting plate
OJ-P-90

Grip mounting plate
OJ-P-90(Needed when using the tensile test jig.)(Needed when using the tensile test jig.)* Digital force gauge RZE series is optional.

* Digital force gauge RZE series is optional.

A vertical and horizontal manual-type small test standdedicated to the RZE series. Capable of measuring up to500N, the gauge mounting part can be moved by turning theoperation wheel.

A small automatic test stand dedicated to the RZE series forboth vertical and horizontal use. This is a reasonable test standthat is suitable for testing small specimens and short strokes.
When used together with the RZE series, such operations arepossible as overload monitoring and automatic stopping orreversing. Please purchase the optional RZE-OP-1 or RZE-OP-2
for cooperation with the RZE series.

Standard Specifications
Max force

500N (50kgf)

Operation mechanism

3mm/wheel rotation

Stroke240mm

Table size

W200 x D120mm (Vertical), W70 x D60mm (Horizontal)

WeightApprox. 16kg (Vertical), Approx. 12kg (Horizontal)

Dimensions

W200 x D150 x H500mm (including the wheel)

Standard Specifications
Max force

500N (50kgf)

Test speed

10 to 200mm/min

Speed changing

Stepless

Stroke150mm

Displacement display

None

Distance between test center and column

49mm

Table size

W200 x D120mm (Vertical), W70 x D60mm (Horizontal)

Drive motor

DC brushless motor

Driving mechanismTrapezoidal screw

Driving mechanism guide system

Sliding bush

WeightApprox. 20kg (Vertical), Approx. 15kg (Horizontal)

Dimensions

W215 x D235 x H380mm

Power supply

100 to 240VAC, single-phase, 0.5A

Horizontal Motorized Stand
MODEL-2252R* Digital force gauge RZE series is optional.

The gauge mount of this device is operated by motor to moveto the right or left. The Z stage can adjust vertically. The work
mount table has slotted holes for adjusting the jig position
back and forth.
Standard Specifications
Max force

500N (50kgf)

Test speed

10 to 80mm/min

Speed changing5 steps or stepless

Stroke150mm

Displacement display

None

Z stage stroke

40mm

Test center to Z stage

39 to 79mm

Table size

W90 x D90mm

Driving mechanism

Trapezoidal screw

Drive motor

DC brushless motor

WeightApprox. 23kg

Dimensions

W518 x D345 x H228mm

Power supply

100 to 240VAC, single-phase, 0.5A

Manual test stand
MODEL-1345

MODEL-1349* Digital force gauge RZE series is optional.

* Digital force gauge RZE series is optional.

The MODEL-1345 is a manual test stand. The gauge mounting
head is moved up and down by a rotary wheel.

The MODEL-1349 is a lever-type test stand exclusively designed
for compressive testing. The gauge mounting head is moved
up and down by a lever.

Standard Specifications
Max force

500N (50kgf)

Standard Specifications

Stroke43mm (1.75mm/rotation)

Max force

500N (50kgf)

Table size

W180 x D100mm

Stroke43mm/140deg

Max span

210mm

Table size

W180×D100mm

WeightApprox. 9kg

Max span

210mm

Dimensions

W200 x D235 x H435mm

WeightApprox. 9kg

Dimensions

W200 x D235 x H435mm

MODEL-1338

MODEL-2254* Digital force gauge RZE series is optional.

* Digital force gauge RZE series is optional.

The MODEL-1338 is a manual test stand suitable for shortstroke measurement. Turning the knob on the side moves
the force gauge up and down. If you rotate the knob, themounting head will pop up by spring. Ideal for measuring
small workpieces with a small amount of movement.
Standard Specifications
Max force

50N (5kgf)

Stroke

20mm

Max span

70mm

WeightApprox. 5kg

Dimensions

W150 x D200 x H260mm

The MODEL-2254 is a manual test stand for measuring thecaulking strength of wire harnesses, and so forth. The work
mounting heads are moved with a lever. Using the RZE series
provides various functions, including printing and storage onPC. The digital gauge of this device is an option, but the gauge
side jig is included as a standard.
Standard Specifications
Max force

500N (50kgf)

Stroke70mm

Chuck width

0 to 10mm

WeightApprox. 15kg

Dimensions

W525 x D200 x H150mm

High Capacity Force gauge
RZE-S-200
The RZE-S-200 is a separate type of the RZE series and caneasily measure forces up to 2000N. Optional jigs are available
for hand measurement. This can be measured more stably
using a motorized stand.
Standard Specifications* The handle, hook and jig are optional.

Capacity

2000N (200kgf)

Minimum display unit

1N (0.1Kgf)

Measurement unit

N, Kgf, lb selectable

Accuracy±0.2%F.S.

Communication output

USB

Analog output±2V/F.S.

Printer output

SANEI thermal printer, MITSUTOYO DIGIMATIC output

Continuous operation time

30 hours after full charge

Power supply

Dedicated Nickel-metal hydride battery
Dedicated AC adapter: MODEL-780 (100 to 240VAC)

WeightApprox. 375g (Display unit only)Standard accessories

Display: RZE-S
Loadcell: MODEL-3200, Cell cable (1.5m)
AC adapter MODEL-780 (USB cable: includes RZ-USB)

Options

Various cables, Software RZ-3000
Thermal printer: BL2-58SNWJC

Separate type Force gauge
RZE-S-8000 series
The RZE-S-8000 series is a force gauge with a separate loadcell
and display. There are few restrictions on the measurement
location, and it is convenient to carry.
Standard Accessories

015B

014B

013B

012B

011B

016B

Standard Specifications
MODEL

RZE-S-8001

RZE-S-8002

RZE-S-8005

RZE-S-8010

RZE-S-8020

RZE-S-8050

RZE-S-8100

Capacity

10N (1kgf)

20N (2kgf)

50N (5kgf)

100N (10kgf)

200N (20kgf)

500N (50kgf)

1000N (100kgf)

Minimum display unit
Measurement unit
Accuracy
Sampling cycle
Display update interval
OutputContinuous operation timePower supplyWeightStandard accessories
Options0.001N (0.1gf)

0.01N (1gf)

0.1N (10gf)
N, kgf, lb selectable±0.2%F.S.

1ms, 5ms, 16ms, 50ms, 125ms, 250ms selectable
1, 2, 5, 10, 20 times/sec selectable
USB, Analog voltage ±2V F.S., Printer, Motorized stand control, External connection hold30 hours after full charge
Dedicated Nickel-metal hydride battery, Dedicated AC adapter: MODEL-780 (100 to 240VAC)
Approx. 375g (Display unit only)
Display unit: RZE-S, Loadcell: S8000 series with Cell cable (3m)Standard accessories 6 pcs, AC adapter: MOOEL-780 (USB cable: includes RZ-USB)
Various cables, Software RZ-3000, Thermal printer: BL2-58SNWJCPortable Torque gauge
RZE-T-20
The RZE-T-20 portable torque gauge is designed for static
torque measurement such as screw tightening/loosening
and torsion. It has a large and fine display that shows bothclockwise and counterclockwise torque, a GO-NG judgment
function, and a battery charge level indicator. Dependingon the object to be measured, the bit or jig at the top canbe changed. Measured data can be stored in the memory oroutput to a connected printer.
Standard Specifications
Capacity

2000mN•m (20kgf•cm)

Minimum display unit

1mN•m (0.01kgf•cm)

Measurement unitmN•m, kgf•cm selectable

Accuracy±1%F.S.

Display update interval

1, 2, 5, 10, 20 times/sec selectable

Sampling cycle

1ms, 5ms, 16ms, 50ms, 125ms, 250ms selectable

Communication output

USB

Analog output±2V/F.S.

Printer output

SANEI thermal printer, MITSUTOYO DIGIMATIC output

Continuous operation time

30 hours after full charge

Power supply

Dedicated Nickel-metal hydride battery
Dedicated AC adapter: MODEL-780 (100 to 240VAC)
Dedicated USB cable: RZ-USB

WeightApprox. 375g (Display unit only)Standard accessories

Display unit RZE-S, Torque gauges
Cell cable (1.5m), Standard accessories 5 pcs
AC adapter: MOOEL-780 (USB cable: includes RZ-USB)

Options

Various cables, Software RZ-3000
Thermal printer: BL2-58SNWJC

RZE-T-100
The RZE-T-100 portable torque gauge is designed for static
torque measurement such as screw tightening/loosening
and torsion. It has a large and fine display that shows bothclockwise and counterclockwise torque, a GO-NG judgment
function, and a battery charge level indicator. Dependingon the object to be measured, the bit or jig at the top canbe changed. Measured data can be stored in the memory oroutput to a connected printer.
Standard Specifications* The handle is optional .

Capacity

10N•m (100kgf•cm)

Minimum display unit

0.001N•m (0.01kgf•cm)

Measurement unitmN•m, kgf•cm selectable

Accuracy±1%F.S.

Display update interval

1, 2, 5, 10, 20 times/sec selectable

Sampling cycle

1ms, 5ms, 16ms, 50ms, 125ms, 250ms selectable

Communication output

USB

Analog output±2V/F.S.

Printer output

SANEI thermal printer, MITSUTOYO DIGIMATIC output

Continuous operation time

30 hours after full charge

Power supply

Dedicated Nickel-metal hydride battery
Dedicated AC adapter: MODEL-780 (100 to 240VAC)
Dedicated USB cable: RZ-USB

WeightApprox. 375g (Display unit only)Standard accessories

Display unit RZE-S, Torque gauges
Cell cable (1.5m), Standard accessories 5 pcs
AC adapter: MOOEL-780 (USB cable: includes RZ-USB)

Options

Various cables, Software RZ-3000
Thermal printer: BL2-58SNWJC

Desktop type simple test stand
MODEL-1308U

MODEL-1309RZE

Grip mounting plate
OJ-P-90(Needed when using the tensile test jig.)* Digital force gauge
RZE series is optional.

* Digital force gauge
RZE series is optional.

A long-stroke automatic test stand dedicated to the RZEseries. This is a reasonable test stand that is suitable for testsof specimens of large size and long test stroke. When used
together with the RZE Series, it is possible to monitor overload
or automatically stop/reverse when the load value set by theRZE Series is reached. To combine this stand with the RZESeries, please use the optionally available RZE-OP-1 or RZEOP-2.

A small stand for force-displacement measurement dedicatedto the RZE series. With a displacement resolution of 0.1mm,
this stand is suitable for long stroke tensile/compression tests.
A force-displacement FS curve can be drawn on a PC screenby the dedicated software for effective data management.

Standard Specifications

Standard Specifications
Max force

1000N (100kgf)

Force display

Depending on Digital force gauge RZE Series

Test speed

10 to 300mm/min

Speed changing

LO: 10mm/min
MD: 100mm/min
HI: 200mm/min Intended setting is available
VR: 10 to 300mm/min

Stroke400mm

Displacement display

Resolution 0.1mm, Max indication ±400mm

Table size

W295 x D175mm

WeightApprox. 35kg

Dimensions

W300 x D340 x H850mm

Power supply

100 to 120V or 200 to 240VAC, single-phase

Max force

1000N (100kgf)

Test speed

5 to 100mm/min

Speed changing

SteplessStroke

400mm

Displacement display

None

Distance between test
center and column

60mm

Table size

W250 x D140mm

Specimen table size

W120 x D90mm, Center M6

Drive motor

DC brushless motor

Driving mechanism

Trapezoidal screw

Driving mechanism
guide system

Linear ball bush

WeightApprox. 28kg

Dimensions

W255 x D300 x H810mm

Data file processing

Data load/batch load

Dedicated format

Power supply

100 to 240VAC, single-phase, 0.5A

Graph overlap display

Number of registration

100

Graph cut sheet print

Cut sheet horizontal: 1 graph

Major functions
Data acquisition

Real time force/displacement acquisition, graph drawing
Data save/batch save

Dedicated format/CSV format

Cut sheet horizontal: 1 graphPrinting

Graph continuous print

Cut sheet vertical: 2 graph
Cut sheet horizontal: 4 graph
Cut sheet vertical: 8 graph

Pick up resistered data
Pick up resistered data
Measurement list display

Number of registration

100/measurement

Representative value detection

Force Max/Min/Ave

All measurement processing

Force Max/Min/Ave

Graph data list display

Time/displacement/force data values

Stand control

Stand start/Stop during test

USB serial conversion

BUFFALO: BSUSRC0605BS (Recommended/option)

Standard Configuration
Tester body, Application software: RZ-3009,
Cable for connecting RZE series, USB cableGrips & Jigs

MODEL-201
Small-size flat chuck

MODEL-203
Flat chuck

MODEL-207-2K
Flat chuck

MODEL-220-50-45
Small double side operating vise

MODEL-221-25/50
Single roller chuck

MODEL-224-P-4/5
Pin vise

MODEL-225
Small loads of wires, etc.

MODEL-226-1/5/10/15
Pantograph chuck

MODEL-227-20/30
All-purpose type

MODEL-228G-10 to 40/228H-10 to 40
Film chuck

MODEL-230-45
Board mounting jig

MODEL-231
All-purpose type

MODEL-232
All-purpose type

CP-U-40/60/80
Round flat type compression tableHS-2 to 8
Shear jigStandard Specifications

G: Rubber H: Sand paper* A joint screw may be required depending on the object.

Capacity

Jaw Size(W×Dmm)

MODEL-201

50N

20 x 15

8*M6 female

Approx.125

All-purpose type

Parallel flat plate individual screwing

MODEL-203

2kN

25 x 20

14*M10 female

Approx.442

All-purpose type

Parallel flat plate individual screwing

MODEL-207-2K

2kN

25 x 25

12*M10 female

Approx.356

All-purpose type

Parallel flat plate individual screwing

MODEL-220-50-45500N

50 x 45

0 to 45

Approx.958

Vise

Right-hand/left-hand thread

MODEL-221-25

200N

W25-

*M6 female

Approx.115

Film/cloth

Spring opening/closing

MODEL-221-50

2kN

W50-M10-P1.5

Approx.376

Film/cloth

Spring opening/closing

MODEL-224-P-4/5

10N-

0 to φ3.2*M6 female

Approx.12

Round bar

For round bar/collet change typeMODEL

Jaw Max
Opng (mm)Mounting screw

Weight(g)Application

Type

MODEL-225

0.5N

25 x 25

3M6 female

Approx.56

Gold wire

Clip type

MODEL-226-1

5N

Tip jaw width 0.5-1

2

M6 female/M10 maleApprox.97

Small parts

Pantograph type

MODEL-226-5100N

Tip jaw width 52M6 female/M10 maleApprox.100

Small parts

Pantograph type

MODEL-226-10100N

Tip jaw width 10

2M6 female/M10 maleApprox.100

Small parts

Pantograph type

MODEL-226-15

100N

Tip jaw width 15

2

M6 female/M10 maleApprox.107

Small parts

Pantograph type

MODEL-227-20

500N

20 x 104*M6 female

Approx.214

All-purpose type

Pantograph type

MODEL-227-30

500N

30 x 104*M6 female

Approx.233

All-purpose type

Pantograph type

MODEL-228-1020NW101

M6 femaleApprox.37

Thin plate

Screwing

MODEL-228-20

20N

W20

1

M6 femaleApprox.49

Thin plate

Screwing

MODEL-228-30

20N

W30

1M6 female

Approx.64

Thin plate

Screwing

MODEL-228-40

20N

W40

1

M6 femaleApprox.77

Thin plate

Screwing

MODEL-230-45--4-Approx.850

Substrate holding

MODEL-231

200N

15 x 103*M6 female

Approx.78

All-purpose typeOpening by spring pressure, closing by screw

MODEL-232

100N

Tip width 3 x D8

1M6 female

Approx.30

All-purpose typeOpening by spring pressure, closing by screw

CP-U-40100N

φ40-M6 female

Approx.100

Upper compression plate

CP-U-60

100N

φ60-

M6 femaleApprox.210

Upper compression plate

CP-U-80

100N

φ80-

M6 femaleApprox.350

Upper compression plate

HS-230N

W2-

M6 femaleApprox.44

Shear jig

HS-450N

W4-

M6 femaleApprox.46

Shear jig

HS-650N

W6-

M6 femaleApprox.48

Shear jig

HS-850N

W8-

M6 femaleApprox.49

Shear jig

OJ-P-90---

M6 femaleApprox.748

Lower tension jig mounting plate

Manufacturer

AIKOH ENGINEERING CO., LTD.
https://www.aikoh.co.jp
Tokyo Sales Office
Nagoya Sales OfficeOsaka Sales Office

TANI Building 4F, 28-6, Kameido 1-chome, Koto-ku, Tokyo 136-0071 Japan
TEL: +81-3-5858-8816 FAX: +81-3-5858-8836
No.4420-ISO 9001
210, Sakuta 2-chome, Nagakute-shi, Aichi 480-1153 Japan
No.E2203-ISO 14001
TEL: +81-561-64-2331 FAX: +81-561-64-2332
15-7, Hishie 2-chome, Higashi Osaka-shi, Osaka 578-0984 Japan
TEL: +81-72-966-9011. FAX:+ 81-72-966-9017

Overseas sales dept.
Head Office / Osaka Factory 15-7, Hishie 2-chome, Higashi Osaka-shi, Osaka 578-0984 Japan
TEL: +81-72-966-9011 FAX: +81-72-966-9017* Specifications are subject to change without notice.

2024032000S