BA00302P/00/EN/18.16
71335667

Products

Solutions

Valid as of software version
04.00.zz

Operating Instructions
Cerabar S PMC71, PMP71,
PMP75
Process pressure measurement

ServicesCerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

1.

Order code: XXXXX-XXXXXX
Ser. no.:XXXXXXXXXXXX
Ext. ord. cd.: XXX.XXXX.XX

Serial number

2. www.endress.com/deviceviewer

Endress+Hauser
Operations App

3.

A0023555

Make sure the document is stored in a safe place such that it is always available whenworking on or with the device.
To avoid danger to individuals or the facility, read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures.
The manufacturer reserves the right to modify technical data without prior notice. Your
Endress+Hauser Sales Center will supply you with current information and updates to these
Instructions.

2

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Table of contents
1

Document information . . . . . . . . . . . . . . 4

1.1
1.2
1.3
1.4
1.5

Document function . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . 5
Terms and abbreviations . . . . . . . . . . . . . . . . . . . . . 6
Turn down calculation . . . . . . . . . . . . . . . . . . . . . . . 7

2

Basic safety instructions . . . . . . . . . . . . . 8

2.1
2.2
2.3
2.4
2.5
2.6

Requirements concerning the staff . . . . . . . . . . . . . 8
Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Hazardous area . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

3

Identification . . . . . . . . . . . . . . . . . . . . . 10

3.1
3.2
3.3
3.4

Product identification . . . . . . . . . . . . . . . . . . . . . . 10
Device designation . . . . . . . . . . . . . . . . . . . . . . . . 10
Scope of delivery . . . . . . . . . . . . . . . . . . . . . . . . . . 12
CE mark, Declaration of Conformity . . . . . . . . . . 12

4

Installation . . . . . . . . . . . . . . . . . . . . . . . 13

4.1
4.2
4.3
4.4
4.5

Incoming acceptance and storage . . . . . . . . . . . . 13
Installation conditions . . . . . . . . . . . . . . . . . . . . . 13
General installation instructions . . . . . . . . . . . . . 14
Installation instructions . . . . . . . . . . . . . . . . . . . . 14
Post-installation check . . . . . . . . . . . . . . . . . . . . . 25

5

Wiring . . . . . . . . . . . . . . . . . . . . . . . . . . . 26

5.1
5.2
5.3
5.4

Connecting the device . . . . . . . . . . . . . . . . . . . . . 26
Connecting the measuring unit . . . . . . . . . . . . . . 27
Overvoltage protection (optional) . . . . . . . . . . . . 28
Post-connection check . . . . . . . . . . . . . . . . . . . . . 28

6Operation. . . . . . . . . . . . . . . . . . . . . . . . . 29

6.1
6.2
6.3
6.4
6.5
6.6
6.7
6.8
6.9

Onsite display (optional) . . . . . . . . . . . . . . . . . . . 29
Operating elements . . . . . . . . . . . . . . . . . . . . . . . 31
FOUNDATION Fieldbus interface . . . . . . . . . . . . 34
Local operation – onsite display connected . . . . 46
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49
HistoROM®/M-DAT (optional) . . . . . . . . . . . . . . 49
Locking/unlocking operation . . . . . . . . . . . . . . . 52
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Factory setting (reset) . . . . . . . . . . . . . . . . . . . . . 53

7

Commissioning . . . . . . . . . . . . . . . . . . . . 56

7.1
7.2
7.3

Configuring messages . . . . . . . . . . . . . . . . . . . . . 56
Function check . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
Commissioning via an FF configuration program . .
56
Selecting the language and measuring mode . . 58
Position adjustment . . . . . . . . . . . . . . . . . . . . . . . 60

7.4
7.5

Endress+Hauser

7.6
7.7
7.8
7.9

Pressure measurement . . . . . . . . . . . . . . . . . . . . . 61
Level measurement . . . . . . . . . . . . . . . . . . . . . . . . 63
Scaling the OUT parameter . . . . . . . . . . . . . . . . . . 67
Configuring event behavior in accordance with
FOUNDATION Fieldbus Specification FF912 Field
Diagnostic Profile . . . . . . . . . . . . . . . . . . . . . . . . . . 68

8Maintenance . . . . . . . . . . . . . . . . . . . . . 79

8.1
8.2

Cleaning instructions . . . . . . . . . . . . . . . . . . . . . . . 79
Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . 79

9Diagnostics and troubleshooting . . . . 80

9.1
9.2
9.3
9.4
9.5
9.6
9.7
9.8
9.9
9.10
9.11
9.12
9.13

Troubleshooting . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
Diagnostic information on local display . . . . . . . . 81
Diagnostic event in the operating tool . . . . . . . . . 82
Diagnostic messages in the DIAGNOSTIC
Transducer Block (TRDDIAG) . . . . . . . . . . . . . . . . 83
Overview of diagnostic events . . . . . . . . . . . . . . . . 86
Response of outputs to errors . . . . . . . . . . . . . . . . 95
Confirming messages . . . . . . . . . . . . . . . . . . . . . . . 96
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Repair of Ex-certified devices . . . . . . . . . . . . . . . . 97
Spare Parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Software history . . . . . . . . . . . . . . . . . . . . . . . . . . . 98

10

Technical data . . . . . . . . . . . . . . . . . . . . 98
Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . 99

3

Document information

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

1

Document information

1.1

Document function

These Operating Instructions contain all the information that is required in various phasesof the life cycle of the device: from product identification, incoming acceptance and storage,
to mounting, connection, operation and commissioning through to troubleshooting,
maintenance and disposal.

1.2

Symbols used

1.2.1

Safety symbols

SymbolMeaning

DANGER
A0011189-DE

WARNING
A0011190-DE

CAUTION
A0011191-DE

NOTICE
A0011192-DE

1.2.2

DANGER!
This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inseriousor fatal injury.
WARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inseriousor fatal injury.
CAUTION!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminoror medium injury.
NOTICE!
This symbol contains information on procedures and other facts which do not result inpersonalinjury.

Electrical symbols

SymbolMeaning

Meaning

Direct current

Alternating current

Direct current and alternating current

Ground connection
A grounded terminal which, as far as
the operator is concerned, is grounded
via a grounding system.

Protective ground connection
A terminal which must be connected
to ground prior to establishing anyother connections.

1.2.3

Symbol)Equipotential connection
A connection that has to be connectedto the plant grounding system: Thismay be a potential equalization line ora star grounding system depending on
national or company codes of practice.

Tool symbols

SymbolMeaning
Allen key
A0011221

Hexagon wrench
A0011222

4

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

1.2.4

Document information

Symbols for certain types of information

SymbolMeaning
Permitted
Indicates procedures, processes or actions that are permitted.
A0011182

Forbidden
Indicates procedures, processes or actions that are forbidden.
A0011184

Tip
Indicates additional information.
A0011193

Reference to documentation
A0028658

Reference to pageA
A0028659

Reference to graphic
A0028660

1. , 2. , 3. …

Series of steps

A0031595

Result of a sequence of actions
A0018343

Visual inspection
A0028673

1.2.5

Symbols in graphics

SymbolMeaning

1, 2, 3, 4, ...

1. , 2. , 3. …

Item numbers
Series of steps

A0031595

A, B, C, D, ...

1.2.6

Views

Symbols at the device

SymbolMeaning
Safety instructions
Observe the safety instructions contained in the associated Operating Instructions.
A0019159

1.3

Registered trademarks

KALREZ, VITON, TEFLON
Registered trademarks of E.I. Du Pont de Nemours & Co., Wilmington, USA
TRI-CLAMP
Registered trademark of Ladish & Co., Inc., Kenosha, USA
FOUNDATIONTM Fieldbus
Registered trademark of the FieldComm Group, Austin, USA
GORE-TEX®
Registered trademarks of W.L. Gore & Associates, Inc., USA

Endress+Hauser

5Document information

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

1.4

Terms and abbreviations

1

2

3

4

p
LRL

0
LRV

URV

URL

MWP

OPL
A0029505

6

Position

Term/Abbreviation

Explanation

1

OPL

The OPL (over pressure limit = sensor overload limit) for the sensors
depends on the lowest-rated element, with regard to pressure, of theselected components, i.e. the process connection must be taken into
consideration in addition to the measuring cell. Also observe pressuretemperature dependency. For the relevant standards and additional notes,
see technical information.
The OPL may be applied for a limited time period.

2

MWP

The MWP (maximum working pressure) for the sensors depends on thelowest-rated element, with regard to pressure, of the selected components,
i.e. the process connection has to be taken into consideration in addition to
the measuring cell. Also observe pressure-temperature dependency. For the
relevant standards and additional notes, see technical information.
The MWP may be applied for an unlimited time.

3

Maximum sensor
measuring range

Range between LRL and URL
This span is the maximum calibratable/adjustable measuring span.

4

Calibrated/Adjusted
measuring span

Range between LRV and URV
Factory setting: 0...URL
Other calibrated spans can be ordered with customised settings.

p-

Pressure-

LRL

Lower range limit-

URL

Upper range limit-

LRV

Lower range value-

URV

Upper range value-

TD

Turn down

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

1.5

Document information

Turn down calculation
1=2

3
URV

LRL LRV

URL
A0029545

Fig. 1:
1
2
3

Calibrated/Adjusted measuring span
Zero-based span
Upper range limit

Example• Calibrated/Adjusted measuring span: 0...5 bar(0...75 psi)
• Lower range value (LRV) = 0 bar• Upper range value (URV) = 5 bar (75 psi)

• Sensor: 10 bar (150 psi)
• Upper range limit (URL) = 10 bar (150 psi)
Turn down (TD):
TD =

TD =

URL
|URV

-

LRV|

10 bar (150 psi)
|5 bar (75 psi)-

0 bar (0 psi)|= 2

In this example, the TD is thus 2:1.
This span is based on the zero point.

Endress+Hauser

7Basic safety instructions

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

2

Basic safety instructions

2.1

Requirements concerning the staff

The personnel for installation, commissioning, diagnostics and maintenance must fulfill thefollowing requirements:• Trained, qualified specialists: must have a relevant qualification for this specific function
and task• Are authorized by the plant owner/operator• Are familiar with federal/national regulations• Before beginning work, the specialist staff must have read and understood the instructionsin the Operating Instructions and supplementary documentation as well as in thecertificates (depending on the application)• Following instructions and basic conditions
The operating personnel must fulfill the following requirements:• Being instructed and authorized according to the requirements of the task by the facility'sowner-operator• Following the instructions in these Operating Instructions

2.2

Designated use

The Cerabar S is a pressure transmitter for measuring pressure and level.

2.2.1

Incorrect use

The manufacturer is not liable for damage caused by improper or non-designated use.
Verification for borderline cases:
For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistance inverifying the corrosion resistance of fluid-wetted materials, but does not accept any
warranty or liability.

2.3

Workplace safety

For work on and with the device:• Wear the required personal protective equipment according to federal/national
regulations.
• Switch off the supply voltage before connecting the device.

2.4

Operational safety

Risk of injury!
‣ Operate the device in proper technical condition and fail-safe condition only.
‣ The operator is responsible for interference-free operation of the device.
Conversions to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers:‣ If, despite this, modifications are required, consult with Endress+Hauser.
RepairTo ensure continued operational safety and reliability,‣ Carry out repairs on the device only if they are expressly permitted.
‣ Observe federal/national regulations pertaining to repair of an electrical device.
‣ Use original spare parts and accessories from Endress+Hauser only.

8

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

2.5

Basic safety instructions

Hazardous area

To eliminate a danger for persons or for the facility when the device is used in the hazardous
area (e.g. explosion protection, pressure vessel safety):• Based on the nameplate, check whether the ordered device is permitted for the intended
use in the hazardous area.
• Observe the specifications in the separate supplementary documentation that is an
integral part of these Instructions.

2.6

Product safety

This measuring device is designed in accordance with good engineering practice to meetstate-of-the- art safety requirements, has been tested, and left the factory in a condition inwhich they are safe to operate. It fulfills general safety requirements and legal requirements.
It also conforms to the EC directives listed in the device-specific EC declaration of conformity.
Endress+Hauser confirms this fact by applying the CE mark.

Endress+Hauser

9Identification

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

3

Identification

3.1

Product identification

The following options are available for identification of the measuring device:• Nameplate specifications• Order code with breakdown of the device features on the delivery note
• Enter serial numbers from nameplates in W@M Device Viewer(www.endress.com/deviceviewer): All information about the measuring device isdisplayed.
For an overview of the technical documentation provided, enter the serial number from thenameplates in the W@M Device Viewer (www.endress.com/deviceviewer).

3.2

Device designation

3.2.1

Nameplates• The MWP (maximum working pressure) is specified on the nameplate. This value refers
to a reference temperature of +20 °C (68°F) and may be applied to the device for an
unlimited time. Observe temperature dependency of the MWP. The pressure values
permitted at higher temperatures can be found in the standards EN 1092-1: 2001 Tab. 18(With regard to their stability-temperature property, the materials 1.4435 and 1.4404 aregrouped together under 13EO in EN 1092-1 Tab. 18. The chemical composition of the twomaterials can be identical.), ASME B 16.5a – 1998 Tab. 2-2.2 F316, ASME B 16.5a –1998 Tab. 2.3.8 N10276, JIS B 2220.
• The test pressure corresponds to the over pressure limit (OPL) of the device = MWP x 1.5
1).
• The Pressure Equipment Directive (2014/68/EU) uses the abbreviation "PS".
The abbreviation "PS" corresponds to the MWP (maximum working pressure) of themeasuring device.
1)

The equation does not apply for PMP71 and PMP75 with a 40 bar (600 psi) or a 100 bar (1500 psi) measuringcell.

Aluminium and stainless steel housing (T14)123Order code:
Ext. order code:

4Ser. no.:5
6

A0016056

Fig. 2:
1
2
3
4
5
6

10

Nameplate
Device name
Order code (for re-orders)
Extended order code (complete)
Technical data
Serial number (for identification)
Address of manufacturer

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Identification

Devices for use in hazardous areas are fitted with an additional nameplate.

1

2
A0021222

Fig. 3:1
2Additional nameplate
Approval-specific information
Document number for safety instructions or drawing number

Devices suitable for oxygen applications or with PVDF process connection are fitted with anadditional nameplate.

1A0022683

Fig. 4:
1

Additional nameplate
Application limits

Hygenic stainless steel housing (T17)

1
2
3
4
5

Order code:
Ext. ord. cd.:
Ser. no.:7

6A0021552

Fig. 5:
1
2
3
4
5
6
7

Nameplate
Device name
Address of manufacturer
Order code (for re-orders)
Extended order code (complete)
Serial number (for identification)
Technical data
Approval-specific information and document number for safety instructions or drawing number

3.2.2

Identifying the sensor type

See parameter "Sensor Meas.Type" in Operating Instruction BA00303P.

Endress+Hauser

11

Identification

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

3.3

Scope of delivery

The scope of delivery comprises:• Cerabar S pressure transmitter• For devices with the "HistoROM/M-DAT" option:
CD-ROM with Endress+Hauser operating program• Optional accessories
Documentation supplied:• Operating Instructions BA00302P and BA00303P are available via the Internet.
See: www.endress.com Download.
• Brief Operating Instructions KA01025P• Fold-out brochure KA00252P• Final inspection report• Additional Safety Instructions with ATEX, IECEx and NEPSI devices• Optional: factory calibration form, test certificates

3.4

CE mark, Declaration of Conformity

The devices are designed to meet state-of-the-art safety requirements, have been tested andleft the factory in a condition in which they are safe to operate. The devices comply with theapplicable standards and regulations as listed in the EC Declaration of Conformity and thus
comply with the statutory requirements of the EC Directives. Endress+Hauser confirms theconformity of the device by affixing to it the CE mark.

12

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4

Installation

4.1

Incoming acceptance and storage

4.1.1

Incoming acceptance

Installation• Check the packaging and the contents for damage.
• Check the shipment, make sure nothing is missing and that the scope of supply matches
your order.

4.1.2
!

Transport

WARNING

Incorrect transportation
Housing, diaphragm and capillaries may become damaged, and there is a risk of injury!
‣ Transport the measuring device to the measuring point in its original packaging or by theprocess connection (with secure transport protection for the diaphragm).
‣ Follow the safety instructions and transport conditions for devices weighing more than
18 kg (39.6 lbs).
‣ Do not use capillaries as a carrying aid for the diaphragm seals.

4.1.3

Storage

The device must be stored in a dry, clean area and protected against impact (EN 837-2).
Storage temperature range:• –40 to +90 °C (–40 to +194 °F)• Onsite display: –40 to +85°C (–40 to +185°F)• Separate housing: –40 to +60°C (–40 to +140°F)

4.2

Installation conditions

4.2.1

Dimensions

For dimensions, please refer to the Technical Information for Cerabar S TI00383P,"Mechanical construction" section.

Endress+Hauser

13Installation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.3

General installation instructions• Devices with a G 1 1/2 thread:
When screwing the device into the tank, the flat seal has to be positioned on the sealing
surfaceof the process connection. To avoid additional strain on the process isolating diaphragm,
the thread should never be sealed with hemp or similar materials.
• Devices with NPT threads:– Wrap Teflon tape around the thread to seal it.
– Tighten the device at the hexagonal bolt only. Do not turn at the housing.
– Do not overtighten the thread when screwing. Max. torque: 20 to 30 Nm (14.75 to
22.13 lbf ft)• For the following process connections a tightening torque of max. 40 Nm (29.50 lbf ft)
is required:– Thread ISO228 G1/2 (Order option "1A" or "1B")– Thread DIN13 M20 x 1.5 (Order option "1N" or "1P")

4.3.1
!

Mounting sensor modules with PVDF thread

WARNING

Risk of damage to process connection!
Risk of injury!
‣ Sensor modules with PVDF thread must be installed with the mounting bracket provided!
!

WARNING

Material fatigue from pressure and temperature!
Risk of injury if parts burst! The thread can become loose if exposed to high pressure andtemperatures.
‣ The integrity of the thread must be checked regularly and the thread may need to be retightened with the maximum tightening torque of 7 Nm (5.16 lbf ft). Teflon tape isrecommended for sealing the ½" NPT thread.

4.4

Installation instructions• Due to the orientation of the Cerabar S, there may be a shift in the zero point, i.e. when thecontainer is empty or partially full, the measured value does not display zero. You cancorrect this zero point shift using the "Zero" key on the electronic insert or externally on thedevice or via the onsite display.  ä 31, Section 6.2.1 "Position of the operating
elements",  ä 32, Section 6.2.2 "Function of the operating elements - onsite display notconnected" and  ä 60, Section 7.5 "Position adjustment".
• For PMP75, please refer to Section 4.4.2 "Installation instructions for devices withdiaphragm seals – PMP75",  ä 17.
• To ensure optimal readability of the onsite display, it is possible to rotate the housing up
to 380°.  ä 24, "Rotating the housing".
• Endress+Hauser offers a mounting bracket for installing on pipes or walls.
 ä 21, Section 4.4.5 "Wall and pipe mounting (optional)".

4.4.1

Installation instructions for devices without diaphragm seals– PMP71, PMC71

NOTICE

Damage to the device!
If a heated Cerabar S is cooled during the cleaning process (e.g. by cold water), a vacuum
develops for a short time, whereby moisture can penetrate the sensor through the pressurecompensation (1).

14Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Installation‣ If this is the case, mount the Cerabar S with the pressure compensation (1) pointing
downwards.

1
FIELD TERMINALS1

SLANIMRET DLEIF1

SLANIMRET DLEIF• Keep the pressure compensation and GORE-TEX® filter (1) free from contamination andwater.
• Cerabar S devices without a diaphragm seal are mounted as per the norms for amanometer (DIN EN 837-2). We recommend the use of shutoff devices and siphons. The
orientation depends on the measuring application.
• Do not clean or touch the process isolating diaphragm with hard or pointed objects.
• The device must be installed as follows in order to comply with the cleanability
requirements of the ASME-BPE (Part SD Cleanability):

IF
T DLE
IMRE
SLAN

FIELD

INALS
TERM

SLANIMRET DLEIF

Pressure measurement in gases➀➁

P01-PMx7xxxx-11-xx-xx-xx-001

Fig. 6:
1
2

Measuring arrangement for pressure measurement in gases
Cerabar S
Shutoff device

Mount Cerabar S with shutoff device above the tapping point so that the condensate can flow
into the process.

Endress+Hauser

15

Installation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Pressure measurement in steams

➀
➃

➁

➁

➂
➀
P01-PMx7xxxx-11-xx-xx-xx-002

Fig. 7:1
2
3
4

Measuring arrangement for pressure measurement in steams
Cerabar S
Shutoff device
U-shaped siphon
Circular siphon• Use siphons for pressure measurement in steam. The siphon reduces the temperature toalmost ambient temperature. Preferably mount the Cerabar S with the siphon below thetapping point.
Advantages:– defined water column only causes minimal/negligible measured errors– only minimal/negligible thermal effects on the device
Mounting above the tapping point is also possible. Pay attention to the maximum
permitted ambient temperature of the transmitter!
• Fill the siphon with liquid before commissioning.
Pressure measurement in liquids➀➁

P01-PMx7xxxx-11-xx-xx-xx-003

Fig. 8:1
2

Measuring arrangement for pressure measurement in liquids
Cerabar S
Shutoff device

Mount Cerabar S with shutoff device below or at the same level as the tapping point.
16

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Installation

Level measurement

P01-PMP75xxx-11-xx-xx-xx-000

Fig. 9:Measuring arrangement for level• Always mount Cerabar S below the lowest measuring point.
• Do not mount the device at the following positions:In the fill flow or at a point in the tankwhich could be affected by pressure pulses from an agitator.
• Do not mount the device in the suction area of a pump.
• The calibration and functional test can be carried out more easily if you mount the device
after a shutoff device.

4.4.2

Installation instructions for devices with diaphragm seals –PMP75• Cerabar S devices with diaphragm seals are screwed in, flanged or clamped, depending on
the type of diaphragm seal.
• Please note that the hydrostatic pressure of the liquid columns in the capillaries can cause
zero point shift. The zero point shift can be corrected.
• Do not clean or touch the process isolating diaphragm of the diaphragm seal with hard or
pointed objects.
• Do not remove process isolating diaphragm protection until shortly before installation.
NOTICE

Improper handling!
Damage to the device!
‣ A diaphragm seal and the pressure transmitter together form a closed, oil-filledcalibrated system. The fill fluid hole is sealed and may not be opened.
‣ When using a mounting bracket, sufficient strain relief must be ensured for thecapillaries in order to prevent the capillary bending down (bending radius 100(3.94 in)).
‣ Please observe the application limits of the diaphragm seal filling oil as detailed in theTechnical Information for Cerabar S TI00383P, "Planning instructions for diaphragm seal
systems" section.
NOTICE

In order to obtain more precise measurement results and to avoid a defect in the device,
mount the capillaries as follows:‣ Vibration-free (in order to avoid additional pressure fluctuations)‣ Not in the vicinity of heating or cooling lines‣ Insulate if the ambient temperature is below or above the reference temperature‣ With a bending radius of  100 mm (3.94 in).
‣ Do not use the capillaries as a carrying aid for the diaphragm seals!

Endress+Hauser

17Installation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Vacuum application
For applications under vacuum, Endress+Hauser recommends mounting the pressuretransmitter below the diaphragm seal. This prevents vacuum loading of the diaphragm sealcaused by the presence of fill fluid in the capillary.
When the pressure transmitter is mounted above the diaphragm seal, the maximum height
difference H1 in accordance with the illustrations below must not be exceeded.

FIELD TERMINALS

H1

A0020472

Fig. 10:Installation above the lower diaphragm seal

The maximum height difference depends on the density of the filling oil and the smallest
ever pressure that is permitted to occur at the diaphragm seal (empty vessel), see illustration
below:
[m]
12.0
1
10.0
2
8.0
3
A 6.0
4
4.0
5

2.0
0.0
50 100 200

300

400

500

600

700

800

900 1000[mbarabs]

B
A0023986-en

Fig. 11:
A
B
1
2
3
4
5

18

Diagram of maximum installation height above the lower diaphragm seal for vacuum applications depending on thepressure at the diaphragm seal
Height difference H1
Pressure at diaphragm seal
Low temperature oil
Vegetable oil
Silicone oil
High-temperature oil
Inert oilEndress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Installation

Mounting with temperature isolator
Endress+Hauser recommends the use of temperature isolators in the event of constantextreme medium temperatures which lead to the maximum permissible electronics
temperature of +85 °C (+185°F) being exceeded. Depending on the filling oil used,diaphragm seal systems with temperature isolators can be used for maximum temperaturesof up to +400 °C (+752 °F). For the temperature application limits, see technical
Information, "Diaphragm seal filling oils" section. To minimize the influence of rising heat,
Endress+Hauser recommends the device be mounted horizontally or with the housing
pointing downwards. The additional installation height also brings about a maximum zero
point shift of 21 mbar (0.315 psi) due to the hydrostatic column in the temperature isolator.
You can correct this zero point shift at the device.
A

B
1

Tp

C
1

Tp

Ta

1
Tp

Ta

Ta

2

2

[°F]

Tp

[°C]

A
+752

+400

+662

+350

+572

+300

+482

+250

+392

+200

+302

+150

+212

+100

+122

+50

+32

0

-58

-50

-148

-100
-60
-76

-40
-40

-20
-4

0
+32

+20
+68

+40
+104

C

+60

B

+80

+140

+176

+100
+212

[°C]
[°F]

Ta
A0032475

Fig. 12:
1
2

Transmitter
Insulation material

Position

Insulation

Ambient temperature (Ta) at transmitter

Maximum process temperature (Tp)A

No insulation

60 °C (140 °F)

400 °C (752 °F) 1)

85 °C (185 °F)

200 °C (392 °F)

80 °C (176 °F)

400 °C (752 °F) )

85 °C (185 °F)

300 °C (572 °F)

67 °C (153 °F)

400 °C (752 °F) )

85 °C (185 °F)

200 °C (392 °F)

B

C

1)

30 mm (1.18 in) Insulation

Maximum insulation

Process temperature: max. 400 °C (752 °F), depending on the diaphragm seal filling oil used

Endress+Hauser

19Installation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.4.3

Seal for flange mounting

NOTICE

Corrupted measurement results.
The seal is not allowed to press against the process isolating diaphragm as this could affectthe measurement result.
‣ Ensure that the seal is not touching the process isolating diaphragm.

1

2

A0017743

Fig. 13:1
Process isolating diaphragm
2
Seal

4.4.4

Installation with heat insulation – PMC71 high temperatureversion and PMP75

The devices must only be insulated up to a certain height. The maximum permitted
insulation height is indicated on the devices and applies to an insulation material with a heat
conductivity  0.04 W/(m x K) and to the maximum permitted ambient and processtemperature( see table below). The data were determined under the most critical application "quiescentair".

Insulationmaterial

TA

TP
P01-PMx7xxxx-11-xx-xx-xx-010

Fig. 14:

Maximum insulation height, here e.g. PMC71 with flange

PMC71

PMP75

Ambient temperature(TA) 70 °C (158°F)

70 °C (158°F)

Process temperature (TP)150 °C (302°F)

 400 °C (752°F)1)

Heat conductivity
insulation material1)20

 0.04 W/(m x K)

Depending on the diaphragm seal filling oil used ( see
Technical Information TI00383P Cerabar S)

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.4.5

Installation

Wall and pipe mounting (optional)

Endress+Hauser offers a mounting bracket for installation on pipes or walls (for pipe
diameters from 1 ¼" to 2").

122 (4.8)
158 (6.22)

ø6 (0.24)

52 (2.05)

86 (3.39)

70 (2.76)

ø42...60 (1.65...2.36)

140 (5.51)
175 (6.89)
A0028493

Engineering unit mm (in)Please note the following when mounting:• Devices with capillary tubes: mount capillaries with a bending radius 100 mm (3.94 in).
• When mounting on a pipe, tighten the nuts on the bracket uniformly with a torque of atleast 5 Nm (3.69 lbs ft).

Endress+Hauser

21

Installation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.4.6

Assembling and mounting the "separate housing" version

6
5
4

7

3
2

1

r ³ 120 (4.72)

mm (in)P01-PMx7xxxx-11-xx-xx-xx-011

Fig. 15:
1
2
3
4
5
6
7

"Separate housing" version
In the "separate housing" version, the sensor is supplied with the process connection and cable ready-fitted.
Cable with connection jackPressure compensation
Plug
Locking screw
Housing fitted with housing adapter, included
Mounting bracket suitable for wall and pipe mounting, included

Assembly and mounting

22

1.

Plug the connector (item 4) into the corresponding connection jack of the cable (item2).

2.

Plug the cable into the housing adapter (item 6).

3.

Tighten the locking screw (item 5).

4.

Mount the housing on a wall or pipe using the mounting bracket (item 7).
When mounting on a pipe, tighten the nuts on the bracket uniformly with a torque ofat least 5 Nm (3.69 lbs ft).
Mount the cable with a bending radius (r) 120 mm (4.72 in).

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.4.7

Installation

PMP71, version prepared for diaphragm seal mount –welding recommendation

1

2

3

ø2.5 (0.1)ø7.95 (0.31)

5 (0.2)

A1

A0028495

Fig. 16:
1
2
3
A1

Version U1: prepared for diaphragm seal mount
Setscrew
Bearing
Hole for fill fluid
See the "Welding recommendation" table below

Engineering unit mm (in)Endress+Hauser recommends welding on the diaphragm seal as follows for the "XSJ Prepared for diaphragm seal mount" version in feature 110 "Process connections" in the ordercode up to, and including, 40 bar (600 psi) sensors: the total welding depth of the fillet weld
is 1 mm (0.04 in) with an outer diameter of 16 mm (0.63 in). Welding is performed
according to the WIG method.
Consecutive seamno.
A1
for sensors
 40 bar (600 psi)

Sketch/welding groove shape,
dimension as per DIN 8551

s1 a0.8

Base material matching

Welding method
DIN EN ISO 24063

Welding
position

Inert gas,
additives

Adapter made of AISI 316L
(1.4435) to be welded todiaphragm seal made of
AISI 316L (1.4435 or 1.4404)

141

PB

Inert gas
Ar/H 95/5
Additive:
ER 316L Si(1.4430)

A0024811

Information on filling
The diaphragm seal must be filled as soon as it has been welded on.
• After welded into the process connection, the sensor assembly must be properly filled witha filling oil and sealed gas-tight with a sealing ball and lock screw.
Once the diaphragm seal has been filled, at the zero point the device display should notexceed 10% of the full scale value of the cell measuring range. The internal pressure of thediaphragm seal must be corrected accordingly.
• Adjustment / calibration:– The device is operational once it has been fully assembled.
– Perform a reset. The device must then be calibrated to the process measuring range as
described in the Operating Instructions.

Endress+Hauser

23

Installation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.4.8

Rotating the housing

The housing can be rotated up to 380° by loosening the Allen screw.
T14

T15

T17

2

3
A0019996

1. T14 housing: Loosen setscrew with a 2 mm (0.08 in) Allen key.
T15 andT17 housing: Loosen setscrew with a 3 mm (0.12 in) Allen key.
2. Rotate housing (max. up to 380 °).
3. Retighten setscrew with 1 Nm (0,74 lbf ft.

4.4.9

Closing the housing cover

NOTICE

Devices with EPDM cover seal - transmitter leakiness!
Mineral-based, animal-based or vegetable-based lubricants cause the EPDM cover seal toswell and the transmitter to become leaky.
‣ The thread is coated at the factory and therefore does not require any lubrication.
NOTICEThe housing cover can no longer be closed.
Damaged thread!
‣ When closing the housing cover, please ensure that the thread of the cover and housing
are free from dirt, e.g. sand.If you feel any resistance when closing the cover, check thethread on both again to ensure that they are free from dirt.
Close cover on a hygenic stainless steel housing (T17)

P01-PMD75xxx-17-xx-xx-xx-000

Fig. 17:

Close cover

The covers for the terminal and electronics compartment are hooked into the casing andclosed with a screw. These screws should be finger-tightened (2 Nm (1.48 lbf ft)) to the stop
to ensure that the covers sit tightly.

24

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

4.5

Installation

Post-installation check

After installing the device, carry out the following checks:• Are all screws firmly tightened?
• Are the housing covers screwed down tight?

Endress+Hauser

25

Wiring

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

5

Wiring

5.1

Connecting the device!

WARNING

Risk of electric shock!
If the operating voltage is > 35 VDC: Dangerous contact voltage at terminals.
‣ In a wet environment, do not open the cover if voltage is present.
!

WARNING

Limitation of electrical safety due to incorrect connection!
• Risk of electric shock and/or explosion in hazardous areas! In a wet environment, do notopen the cover if voltage is present.
• When using the measuring device in hazardous areas, installation must comply with thecorresponding national standards and regulations and the Safety Instructions or
Installation or Control Drawings.
• Devices with integrated overvoltage protection must be grounded.
• Protective circuits against reverse polarity, HF influences and overvoltage peaks are
installed.
• The supply voltage must match the supply voltage on the nameplate. ( ä 10,
Section 3.2.1 "Nameplates".)• Switch off the supply voltage before connecting the device.
• Remove the housing cover of the terminal compartment.
• Guide the cable through the gland. For cable specifications,  ä 27, Section 5.2.4.
• Connect the device in accordance with the following diagram.
• Screw down the housing cover.
• Switch on the supply voltage.

➀

➃

FF

FF

➁

➄
FF

FF

➂
P01-xMx7xxxx-04-xx-xx-xx-009

Fig. 18:
1
2
3
4
5

26

Electrical connection of FOUNDATION Fieldbus
Please refer also to Section 5.2.1 "Supply voltage",  ä 27.
Housing
Internal ground terminal
External ground terminal
Supply voltage, for version in non-hazardous area = 9 to 32 V DC
Devices with integrated overvoltage protection are labeled OVP (overvoltage protection) here.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

5.1.1

WiringConnecting devices with 7/8" plug

PIN assignment for 7/8" connector

1

3

2

4

PIN

Meaning

PIN

Meaning

1

Signal –2

Signal +3

Shield4

Not assigned

A0011176

5.2

Connecting the measuring unit

For further information on the network structure and grounding and for further bus systemcomponents such as bus cables, see the relevant documentation, e.g. Operating Instructions
BA00013S "FOUNDATION Fieldbus Overview" and the FOUNDATION Fieldbus Guideline.

5.2.1

Supply voltage

Version for non-hazardous area: 9 to 32 V DC!

WARNING

Supply voltage might be connected!
Risk of electric shock and/or explosion!
‣ When using the measuring device in hazardous areas, installation must comply with thecorresponding national standards and regulations and the Safety Instructions or
Installation or Control Drawings.
‣ All explosion protection data are given in separate documentation which is available
upon request. The Ex documentation is supplied as standard with all devices approved foruse in explosion hazardous areas.

5.2.2

Current consumption

15.5 mA ±1 mA, switch-on current corresponds to IEC 61158-2, Clause 21.

5.2.3

Terminals• Supply voltage and internal ground terminal: 0.5 to 2.5 mm2 (20 to 14 AWG)
• External ground terminal: 0.5 to 4 mm2 (20 to 12 AWG)

5.2.4

Cable specification• Use a twisted, shielded two-wire cable, preferably cable type A.
• Terminals for wire cross-sections: 0.5 to 2.5 mm2 (20 to 14 AWG)
• Outer cable diameter: 5 to 9 mm (0.2 to 0.35 in)
For further information on the cable specifications, see Operating Instructions BA00013S"FOUNDATION Fieldbus Overview", FOUNDATION Fieldbus Guideline and IEC 61158-2(MBP).

5.2.5

Grounding and shielding

Cerabar S must be grounded, for example by means of the external ground terminal.
Different grounding and shielding installation methods are available for FOUNDATION
Fieldbus networks such as:• Isolated installation (see also IEC 61158-2)
Endress+Hauser

27

Wiring

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus• Installation with multiple grounding• Capacitive installation

5.3

Overvoltage protection (optional)

NOTICE

Device could be destroyed!
Devices with integrated overvoltage protection must be earthed.
Devices showing version "M" in feature 100 "Additional options 1" or feature 110 "Additional
options 2" in the order code are equipped with overvoltage protection (see also Technical
Information TI00383P "Ordering information").
• Overvoltage protection:– Nominal functioning DC voltage: 600 V– Nominal discharge current: 10 kA• Surge current check î = 20 kA as per DIN EN 60079-14: 8/20 s satisfied• Arrester AC current check I = 10 A satisfied

5.4

Post-connection check

Perform the following checks after completing electrical installation of the device:• Does the supply voltage match the specifications on the nameplate?
• Is the device connected as per Section 5.1?
• Are all screws firmly tightened?
• Are the housing covers screwed down tight?
As soon as voltage is applied to the device, the green LED on the electronic insert lights up
for a few seconds or the connected onsite display lights up.

28

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6

Operation

Operation

Feature 20 "Output; operation" in the order code provides you with information on theoperating options available to you.
Versions in the order code

OperationP

FOUNDATION Fieldbus; external operation, LCD

Via onsite display or 1 key on the exterior of the device

Q

FOUNDATION Fieldbus; internal operation, LCD

Via onsite display or 1 key on the inside of the device

R

FOUNDATION Fieldbus; internal operation

Without onsite display, 1 key on the inside of thedevice6.1

Onsite display (optional)

A 4-line liquid crystal display (LCD) is used for display and operation. The onsite display
shows measured values, fault messages and notice messages.
The display of the device can be turned in 90° steps.
Depending on the orientation of the device, this makes it easy to operate the device and readthe measured value.
Functions:• 8-digit measured value display including sign and decimal point, unit display• Bar graph as graphic display of the current pressure measured value in relation to the set
pressure range in the Pressure Transducer Block. The pressure range is set by means of the
SCALE_IN parameter.
• Easy and complete menu guidance by dividing the parameters into several levels andgroups.
• Menu guidance
The onsite display is available in English. Needless to say, the device can also be operated
in 6 languages (de, en, fr, es, jp, ch) via the DTM or EDD. The FieldCare program is an E+H
DTM operating tool and can be acquired from endress.com.
• Each parameter has a 3-digit ID to aid navigation.
• Option of configuring the display according to individual requirements and preferences,such as language, alternating display, contrast setting, display of other measured values
such as sensor temperature• Comprehensive diagnostic functions (fault and warning message, maximum indicator,etc.)• Rapid and safe commissioning using Quick Setup menus.

Endress+Hauser

29

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Parameter
Identification
numberMeasured value display
Function name

Value

Header line
Main line
Information
line
–

+

E

Unit
SymbolBargraph

Editing modes
Operating keys
Selection
options

Value thatcan be edited

Current measured value
P01-xxxxxxxx-07-xx-xx-en-011

The following table illustrates the symbols that can appear on the onsite display. Foursymbols can occur at one time.
SymbolMeaning
Alarm symbol– Symbol flashing: warning, device continues measuring.
– Symbol permanently lit: error, device does not continue measuring.
Note: The alarm symbol may overlie the tendency symbol.
Lock symbol
The operation of the device is locked. For information on unlocking the device, ä 52, Section 6.7 "Locking/unlocking operation".
Communication symbol
Data transfer via communication
Simulation symbol
Simulation mode is activated. DIP switch 2 for simulation is set to "On".
See also Section 6.2.1 "Position of the operating elements"  ä 31, Page 53"Simulation".
Tendency symbol (increasing)
The primary value of the Pressure Transducer Block is increasing.
Tendency symbol (decreasing)
The primary value of the Pressure Transducer Block is decreasing.
Tendency symbol (constant)
The primary value of the Pressure Transducer Block has remained constant over the
past few minutes.

30

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Operation

6.2

Operating elements

6.2.1

Position of the operating elements

The operating keys of the housing T14 (aluminum or stainless steel) are located either
outside of the housing, under the protection cap or upon the electronic insert. The operatingkeys of the hygienic stainless steel housing T17 are located inside the housing upon theelectronic insert. In addition, there are three operating keys on the optional onsite display.

Sim.

➄➅
on

➀

➀
off

1

➁

➂

2

➃

0%
Zero
Simulation

Sensor

0%
Zero

on
off

HW

Histo ROM

2

PC

1

DisplayP01-xMD7xxxx-19-xx-xx-xx-075

Fig. 19:
1

Operating key external, under the protective flap
Operating key for position adjustment (zero point
correction) and total reset

P01-xxxxxxxx-19-xx-xx-xx-106

Fig. 20:
1
2
3
4
5
6

Endress+Hauser

Operating key and operating elements, internal
Green LED to indicate value is accepted
Operating key for position adjustment (zero point
correction) and total reset
Slot for optional display
Slot for optional HistoROM®/M-DAT
DIP switch for locking/unlocking parameters relevant to
the measured value
DIP switch for simulation mode31

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.2.2

Function of the operating elements onsite display not connected

Key(s)

Meaning

0%
Zero

Sim.

P02-xxxxxxxx-19-xx-xx-xx-107

on

1

2

off

P01-xxxxxxxx-19-xx-xx-xx-134– Position adjustment (zero point correction): press key for at least 3 seconds. The
LED on the electronic insert lights up briefly if the pressure applied has been
accepted for position adjustment.
 See also the following section "Performing POSITION ADJUSTMENT on site."– Total reset: press key for at least 12 seconds. The LED on the electronic insert
lights up briefly if a reset is being carried out.
– DIP switch 1: for locking/unlocking parameters relevant to the measured value.
Factory setting: off (unlocked)
See also  ä 52, Section 6.7 "Locking/unlocking operation".
– DIP switch 2: for simulation mode
Factory setting: off (simulation mode off)See also  ä 53, Section 6.8 "Simulation".

Performing position adjustment on site
• Operation must be unlocked.  ä 52, Section 6.7 "Locking/unlocking operation".
• The device is configured for the Pressure measuring mode as standard.
– Operation via FF configuration program: In the Pressure Transducer Block, change themeasuring mode by means of the PRIMARY_VALUE_TYPE and LINEARIZATION
parameters.
– Operation via digital communication: change the measuring mode by means of the
MEASURING MODE parameter.
– You can change the measuring mode by means of the MEASURING MODE parameter.
See  ä 58, Section 7.4 "Selecting the language and measuring mode".
• The pressure applied must be within the nominal pressure limits of the sensor. See
information on the nameplate.
Perform position adjustment:

32

1.

Pressure is present at device.

2.

Press key for at least 3 seconds.

3.

If the LED on the electronic insert lights up briefly, the pressure applied has been
accepted for position adjustment.
If the LED does not light up, the pressure applied was not accepted. Observe the input
limits. For error messages,  ä 81, Section 9.2 "Diagnostic information on local
display".

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.2.3

Operation

Function of the operating elements – onsite display connected

Key(s)

MeaningO
SF– Navigate upwards in the picklist– Edit numerical values or characters within a function– Navigate downwards in the picklist– Edit numerical values or characters within a function– Confirm entry– Go to next item

O and F
S and F

O and S

Contrast setting of onsite display: increase

Contrast setting of onsite display: reduce

ESC functions:– Exit the editing mode without saving the altered value– You are in the menu within a function group: the first time you press the keys
simultaneously, you go back one parameter in the function group. Everysubsequent time you press the keys simultaneously, you go up one level in themenu.
– You are in the menu on a selection level: every time you press the keys
simultaneously, you go up one level in the menu.

Sim.

Note: For the terms function group, level, selection level,  ä 46, Section 6.4.1

on

1

2

– DIP switch 1: for locking/unlocking parameters relevant to the measured value.
Factory setting: off (unlocked)– DIP switch 2: for the simulation mode
Factory setting: off (simulation mode off)

off

P01-xxxxxxxx-19-xx-xx-xx-134

Endress+Hauser

33

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.3

FOUNDATION Fieldbus interface

6.3.1

System architecture

The following diagram shows two typical examples of a FOUNDATION Fieldbus networkwith the associated components.

Industrial Network

FF-HSE
BT

FF-H1

LD

SB

BT

PS

BT
PS

FF-H1
BT

P01-xxxxxxxx-02-xx-xx-xx-001

Fig. 21:

FOUNDATION Fieldbus system architecture with associated components

FF-HSE: High Speed Ethernet, FF-H1: FOUNDATION Fieldbus-H1, LD: Linking Device FF-HSE/FF-H1, PS: Bus Power Supply, SB: Safety
Barrier, BT: Bus Terminator
The system can be connected in the following ways:– A linking device makes the connection to higher-order fieldbus levels (e.g. High Speed Ethernet (HSE)) possible.
– An FF-H1 connecting card is needed for direct connection to a process control system.

Further information on FOUNDATION Fieldbus can be found in Operating Instructions
BA00013S "FOUNDATION Fieldbus Overview, Installation and Commissioning Guidelines",
the FOUNDATION Fieldbus Specification or on the Internet at "http://www. fieldbus.org".

34

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.3.2

Operation

Number of devices• Endress+Hauser Cerabar S devices meet the requirements of the FISCO model.
• Due to the low current consumption, the following can be operated at one bus segment
when
installation is performed according to FISCO:
Up to HW version 1.10:– Up to 7 Cerabar S devices for Ex ia, CSA and FM IS applications– Up to 25 Cerabar S devices in all other applications, e.g. in non hazardous areas, Ex nA
etc.
As of HW version 02.00:– Up to 6 Cerabar S devices for Ex ia, CSA and FM IS applications– Up to 24 Cerabar S devices in all other applications, e.g. in non hazardous areas, Ex nA
etc.
The maximum number of measuring devices at one bus segment is defined by their current
consumption, the performance of the bus coupler, and the required bus length.
As of hardware version 1.10, you will find a label in the device on the electronic insert.

6.3.3

Operation

You can obtain special configuration and operating programs from various manufacturers
for the configuration, such as the FieldCare operating program from Endress+Hauser ä 49, Section 6.5 "FieldCare". These configuration programs make it possible to configure
FF functions and all the device-specific parameters. The predefined function blocks allow
uniform access to all the network and device data.

6.3.4

Network configuration

You require the following to configure a device and integrate it into an FF network:• An FF configuration program• The Cff file (Common File Format: *.cff, *.fhx)• The device description (Device Description: *.sym, *.ffo, *.sy5, *.ff5)
Predefined standard DDs, which can be obtained from FOUNDATION Fieldbus, are available
for the basic functions of measuring devices. You require the device-specific DD to be able toaccess all the functions.
The files for Cerabar S can be acquired as follows:• Internet Endress+Hauser: http://www.de.endress.com  Search for FOUNDATION
Fieldbus• Internet FOUNDATION Fieldbus: http://www.fieldbus.org• On CD-ROM from Endress+Hauser, order number: 56003896
The device is integrated into the FF network as follows:• Start the FF configuration program.
• Download the Cff and device description files (ffo, *.sym, *.cff or *.fhx files) to the system.
• Configure the interface, see Note.
• Configure the device for the measuring task and for the FF system.
• For more in-depth information on integrating the device into the FF system, see thedescription for the configuration software used.
• When integrating the field devices into the FF system, make sure you are using the rightfiles. You can read out the required version by means of the DEV_REV and DD_REV
parameters in the Resource Block.

Endress+Hauser

35

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.3.5

Device identification and addressing

FOUNDATION Fieldbus identifies the device using its ID code and automatically assigns it asuitable field address. The identity code cannot be changed.
The device appears in the network display once you have started the FF configuration
program and integrated the device into the network. The blocks available are displayed
under the device name.
If the device description has not yet been loaded, the blocks report "Unknown" or "(UNK)".
Cerabar S reports as follows:
–

➀
–

➁

EH_CerabarS-XXXXXXXXXXXXXXXX
RS_XXXXXXXXXXX (RB2)
TRD1_XXXXXXXXXXX (PCD)SERVICE_XXXXXXXXXXX (SERVICE)DIAGNOSTIC_XXXXXXXXXXX (DIAGNOSTIC)DISPLAY_XXXXXXXXXXX (DISP)
AI1_XXXXXXXXXXX (AI)AI2_XXXXXXXXXXX (AI)DI_XXXXXXXXXXX (DI)
DO_XXXXXXXXXXX (DO)PID_XXXXXXXXXXX (PID)
ARTH_XXXXXXXXXXX (ARB)
CHAR_XXXXXXXXXXX (SCB)
ISEL_XXXXXXXXXXX (ISB)
INTG_XXXXXXXXXXX (ITB)
AALM_XXXXXXXXXXX (AALB)
P01-PMx7xxxx-05-xx-xx-xx-010

Fig. 22:
1
2

Typical Cerabar S display in a configuration program after the connection has been established
Device name
Serial number

6.3.6

Cerabar S block modelWith FOUNDATION Fieldbus, all the device parameters are categorized according to theirfunctional properties and task and are generally assigned to three different blocks.
A FOUNDATION Fieldbus device has the following block types:• A Resource Block (device block):
This block contains all the device-specific features of the device.
• One or more Transducer Blocks
A Transducer Block contains all the measuring and device-specific parameters of thedevice. The measuring principles, such as pressure, are mapped in the Transducer Blocks.
• One or more function blocks:
Function blocks contain the automation functions of the device. A distinction is madebetween different function blocks such as the Analog Input Block or Proportional Integral
Differential Block (PID). Each of these function blocks is used to execute different
application functions.
The function blocks can be connected by means of an FF configuration program, dependingon the automation task. The device thus takes on simple control functions, thereby relieving
the workload on the higher-order process control system.

36

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Operation

Cerabar S has the following blocks:• Resource Block (device block)• 4 Transducer Blocks– Pressure Transducer Block (TRD)
This Block supplies the output variables PRIMARY_VALUE and SECONDARY_VALUE. It
contains all the parameters to configure the measuring device for the measuring tasksuch as measuring mode selection, linearization function and unit selection.
– Service Transducer BlockThis Block supplies the output variables COUNTER P_PMAX, PRESSURE_1_ MAX_RESETABLE and PRESSURE_1_AFTER_DAMPING. It also includes all the counters formeasuring range overshoot/undershoot for pressure and temperature, minimum andmaximum measured values for pressure and temperature and the HistoROM function.
– Display Transducer BlockThis Block does not return any output variables. It contains all the parameters forconfiguring the onsite display such as DISPLAY_CONTRAST.
– Diagnostic Transducer BlockThis Block does not return any output variables. It contains– the simulation function for the Pressure Transducer Block– parameters to configure the alarm response– parameters to set the user limits for pressure and temperature.
• 8 function blocks– 2 Analog Input Blocks (AI)– Discrete Output Block (DO)– Discrete Input Block (DI)– PID Block (PID)– Arithmetic Block (ARB)– Signal Characterizer Block (SCB)– Input Selector Block (ISB)– Analog Alarm Block (AALB)– Integrator Block (IT)In addition to the pre-instantiated blocks already mentioned, the following blocks can also
be instantiated:• 3 Analog Input Blocks (AI)
• 1 Discrete Output Block (DO)
• 1 Discrete Input Block (DI)
• 1 PID Block (PID)• 1 Arithmetic Block (ARB)• 1 Signal Characterizer Block (SCB)• 1 Input Selector Block (ISB)• 1 Analog Alarm Block (AALB)• 1 Integrator Block (IT)A total of 20 blocks can be instanced in Cerabar S altogether, including the blocks alreadyinstanced. For instancing blocks, see the appropriate Operating Instructions of theconfiguration program used.
Endress+Hauser Guideline BA00062S.
The guideline provides an overview of the standard function blocks that are described in
FOUNDATION Fieldbus Specifications FF 890 - 894.
It is designed to help operators use the blocks implemented in the Endress+Hauser fielddevices.

Endress+Hauser

37

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Block configuration when device is delivered
The block model shown below illustrates the block configuration when the device isdelivered.
Measured variable

SensorSignal evaluation

Pressure
Transducer Block

Display
with scaling

Primary value
Sensore temperature

Service
Transducer BlockResource Block

Analog Input Block 1
CHANNEL = 1
L_TYPE = Direct

Analog Input Block 2
CHANNEL = 2
L_TYPE = Direct

Display
Transducer Block
Arithmetic BlockPID Block

Input Selector
Block

Discrete Input
Block

Discrete Output
Block

SignalCharacterizer
Block

Analog Alarm
Block

Integrator
Block

Diagnostic
Transducer Block

P01-PMx7xxxx-02-xx-xx-en-006

Fig. 23:

Block configuration when device is delivered

The Pressure Transducer Block supplies the Primary Value and the sensor temperature(secondary value). The Primary Value and Secondary Value are each transferred to an
Analog Input Block by means of the CHANNEL parameter (see also the following section).
The Discrete Output, Discrete Input, PID, Arithmetic, Signal Characterizer, Input Selector,
Integrator and Analog Alarm Block are not connected in the as-delivered configuration.
!

CAUTION

Note Dependencies when setting parameters!
‣ Please note that the links between the blocks are deleted and the FF parameters are resetto the default values following a reset by means of the RESTART parameter in the
Resource Block, "Default" option.

38

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.3.7

Operation

Assignment of Transducer Blocks (CHANNEL)

Settings for the Analog Input Block
Process variable

Transducer Block

Parameter name

CHANNEL parameter inthe Analog Input Block

Primary Value, a pressure orlevel value depending on themeasuring mode1)

Pressure Transducer
Block

PRIMARY_VALUE

1MEASURED_TEMPERA
TURE/TEMP. SENSOR

2

Secondary Value(sensor temperature)2)
Pressure after damping

Service Transducer Block PRESSURE_1_AFTER_D 3
AMPING/PRESSURE

Maximum measured pressure

PRESSURE_1_MAX_RE
STABLE/MAX. MEAS.
PRESS.

4

Overshoot counter for
maximum set user limit forpressure

COUNTER: P > Pmax

5

Parameter name

CHANNEL parameter inthe Discrete Output
Block

1)

Factory setting for Analog Input Block 1

2)

Factory setting for Analog Input Block 2

Settings for the Discrete Output Block
Process variable

Transducer Block

Overshoot counter for
maximum set user limit forpressure1)

Service Transducer Block COUNTER: P > Pmax

1)

Endress+Hauser

1

Factory setting

39

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Discrete Input Block settings
Alarm conditions

Transducer Block

Parameter name

General device error

1

Configuration error

2

Sensor overpressure

3

Sensor underpressure

4

Sensor overtemperature

5

Sensor undertemperature

6

Process isolating diaphragm broken
Electronic overtemperature

40

Parameter CHANNEL,
Discrete Input Block

7
Diagnostic
Transducer BlockDIAGNOSTIC_CODE

8

Electronic undertemperature

9

Temperature transmitter override

10

Pressure transmitter underride

11

Pmin PROCESS underrun

12

Pmax PROCESS overrun

13

Tmin PROCESS underrun

14

Tmax PROCESS overrun

15

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.3.8

Operation

Index tables of Endress+Hauser parameters

The following tables list the manufacturer-specific device parameters for the Resource Block,the Transducer Blocks and the Analog Input Blocks. For the FF parameters, see either the
FF specification or Operating Instructions BA00303P "Description of Device Functions,
Cerabar S/Deltabar S/Deltapilot S". These parameters are not displayed in the block view in
FieldCare (exception: Analog Input Blocks).
General explanatory remarks
Data type• DS: data structure, contains data types such as Unsigned8, Octet String etc.
• Bit enumerated• Float: IEEE 754 format• Visible String: ASCII coded• Unsigned:– Unsigned8: value range = 0 to 255– Unsigned16: value range = 0 to 65535
Storage Class• D: dynamic parameter• N: nonvolatile parameter• S: static parameter
If this is a write parameter, the MODE_BLK column indicates the block mode in which theparameter can be written. Some parameters can only be written in the OOS block mode.
The "Reset codes" column indicates which reset codes reset the parameter.

Resource Block
Parameter name,option "Symbolic name"Parameter name,option "Label"

Index Data type

Size
Storage Read Write MODE_BLK Reset(Byte) Class
codes

ENP_VERSION
DEVICE_TAGSERIAL_NUMBER
ORDER_CODE
FIRMWARE_VERSION
SW_LOCKSTATUS_LOCKING
HARDWARE_REVISION
FF_COMM_VERSIONBLOCK_ERR_DESC_1
DEVICE_DIALOG
ELECTRONIC_SERIAL_NUMBER
PROCESS_CONNECTION_TYPE
MAT_PROC_CONN_POS
MAT_PROC_CONN_NEG
SEAL_TYPESCI_OCTET_STRING
MS_RESOURCE_DIRECTORY

ENP version
Device tag
Serial number
Order code
Firmware version
Insert PIN no.
Status locking
Hardware rev.
FF comm. version
Block Error desc.
Device dialog
Electr. serial no.
Proc. conn. type
Mat. proc. conn. +Mat. proc. conn. Seal type
SCI_OCTET_STR
RESOURCE DIRECTORY

44
45
46
47
48
49
50
74
75
76
77
78
79
80
81
82
83
84

16
32
16
32
16
2
2
16
16
4
1
16
2
2
2
2
40
20x2

1)

Visible String
Visible String
Visible String
Visible String
Visible String
Unsigned16
Unsigned16
Visible String
Visible String
Bit enumerated
Unsigned8
Visible String
Unsigned16
Unsigned16
Unsigned16
Unsigned16
Visible String
Unsigned16

S
S
S
S
S
S
D
S
S
D
D
S
S
S
S
S
S
S

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

x1)
x)
x)

AUTO, OOS
AUTO, OOS
AUTO, OOSx

AUTO, OOS

7864, 333

x
x
x
x
x

AUTO, OOS
AUTO, OOS
AUTO, OOS
AUTO, OOS
AUTO, OOS

7864, 333
7864, 333
7864, 333
7864, 333

Can be written with service code

Endress+Hauser

41

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Pressure Transducer Block
Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage ReadClassMEASURED_TEMPERATURE
MEASURED_TEMPERATURE_UNIT
DEVICE_DIALOG
SW_LOCK

TemperatureTemp. eng. unit
Device dialog
Insert PIN no.

32
33
34
35

DS-65
Unsigned16
Unsigned8
Unsigned16

5
2
1
2

D
S
D
S

x
x
x
x

STATUS_LOCKINGLINEARIZATIONSCALE_IN
SCALE_OUTDAMPING_VALUE
ZERO_POSITION_ADJUSTPOSITION_INPUT_VALUE

Status locking
Linearization
Scale In
Scale Out
Damping value
Pos. zero adjust
Pos. input value

36
37
38
39
40
41
42

Unsigned16
Unsigned8
DS-68
DS-68
Float
Unsigned8
Float

2
2
11
11
4
1
4

D
S
S
S
S
D
S

CALIBRATION_OFFSET

Calib. offset

43

Float

4

CUSTOMER_UNIT_PRESSURE

Customer unit P

44

CUSTOMER_FACTOR_UNIT_PRESS
LOW_TRIM_MEASURED
HIGH_TRIM_MEASUREDLEVEL_MODE
LINEAR_MEASURAND
LINEARIZED_MEASURAND
COMBINED_MEASURAND
DENSITY_UNIT
HEIGHT_UNIT
CUSTOMER_HEIGHT_UNIT

Cust. unit. fact. P
Lo trim measured
Hi trim measured
Level mode
Lin. measurandLin. measurandComb. measurandDensity unit
Height unit
Customer unit H

CUSTOMER_UNIT_FACTOR_HEIGHTVOLUME_UNIT
CUSTOMER_UNIT_VOLUME

Write MODE_BLK

Reset codes

x

OOS

x

AUTO, OOS,
MAN

7864, 333

x
x
x
x
x
x
x

x
x
x
x
x
x

OOS
OOS
OOS
OOS
OOS
OOS

7864, 333
7864, 333
7864, 333
7864, 333

S

x

x

OOS

Visible String 8

S

x

x

45
46
47
48
49
50
51
52
53
54

Float
Float
Float
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned16
Unsigned16
Visible String

4
4
4
1
1
1
1
2
2
8

S
S
S
S
S
S
S
S
S
S

x
x
x
x
x
x
x
x
x
x

x

AUTO, OOS,
MAN
OOS

x
x
x
x
x
x
x

Cust. unit. fact. H
Unit volume
Customer unit V

55
56
57

Float
4
Unsigned16 2
Visible String 8

S
S
S

x
x
x

x
x
x

CUSTOMER_UNIT_FACTOR_VOLUMEMASS_UNIT
CUSTOMER_UNIT_MASS

Cust. unit. fact. V
Unit mass
Customer unit M

58
59
60

Float
4
Unsigned16 2
Visible String 8

S
S
S

x
x
x

x
x
x

CUSTOMER_UNIT_FACTOR_MASSCALIBRATION_MODE
ADJUST_DENSITY
ZERO_POSITIONEMPTY_CALIBRATIONFULL_CALIBRATIONTANK_VOLUMETANK_HEIGHT
HUNDRED_PERCENT_VALUELEVEL_MINLEVEL_MAX
PROCESS_DENSITYLINEARIZATION_TABLE_SELECTION

Cust. unit. fact. M
Calibration mode
Adjust density
Zero position
Empty calib. level easy
Full calib. level easy
Tank volume
Tank height
100% point
Level Min.
Level Max.
Process density
Table selection

61
62
63
64
65
66
67
68
69
70
71
72
73

Float
Unsigned8
Float
Float
Float
Float
Float
Float
Float
Float
Float
Float
Unsigned8

8
1
4
4
4
4
4
4
4
4
4
4
1

S
S
S
S
S
S
S
S
S
S
S
S
S

x
x
x
x
x
x
x
x
x
x
x
x
x

x
x
x
x
x
x
x
x
x
x
x
x
x

LINEARIZATION_EDIT_MODE

Edit table

74

Unsigned8

1

S

x

x

LINEARIZATION_TABLE_PRE_EDIT

Editor table

75

Unsigned8

1

D

x

x

LINEARIZATION_TABLE_INDEXLine numb:76

Unsigned8

1

D

x

x

LINEARIZATION_TABLE_X_VALUE

X-value:

77

Float

4

S

x

x

LINEARIZATION_TABLE_Y_VALUE

Y-value:

78

Float

4

S

x

x

LINEARIZATION_TABLE_POST_EDIT
LINEARIZATION_TABLE_POST_VIEW

Table editor
Measuring table

79
80

Unsigned8
Unsigned8

1
1

D
D

x
x

x
x

LEVEL_TANK_DESCRIPTION

Tank description

81

Visible String 32

S

x

x

SENSOR_PRESSURE
PRESSURE
LEVEL_BEFORE_LINEARISATION

Sensor pressure
Pressure
Level before lin

82
83
84

Float
Float
Float

D
D
D

x
x
x

42

4
4
4

OOS
OOS
OOS
OOS
OOS
OOS
AUTO, OOS,
MAN
OOS
OOS
AUTO, OOS,
MAN
OOS
OOS
AUTO, OOS,
MAN
OOS
OOS
OOS
OOS
OOS
OOS
OOS
OOS
OOS
OOS
OOS
OOS
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
OOS
AUTO, OOS,MAN
AUTO, OOS,
MAN

7864, 333,
2509
7864, 333,
2509
7864
7864
2509
2509
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864
7864
7864, 333
7864
7864
7864, 333
7864
7864
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864, 333
7864

7864
7864

7864

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Operation

Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage ReadClass

SENSOR_MEAS_TYPELEVEL_SELECTION
HEIGHT_UNIT_EASYOUTPUT_UNIT_EASYCALIBRATION_MODE_EASY

Sensor meas. typeLevel mode
Height unit
Output unit
Calibration mode level
easy
Density unit level easy
Adjust density level easy
Full calib. level easy
Full calib. level easy
Process density level easy
Meas. level easy
Full calib. level easy
Full calib. level easy
Full pressure
Empty pressure

85
86
87
88
89

Unsigned16
Unsigned8
Unsigned16
Unsigned16
Unsigned8

2
1
2
2
1

D
S
S
S
S

x
x
x
x
x

90
91
92
93
94
95
96
97
98
99

Unsigned16
Float
Float
Float
Float
Float
Float
Float
Float
Float

2
4
4
4
4
4
4
4
4
4

S
S
S
S
D
D
S
S
S
S

x
x
x
x
x
x
x
x
x
x

DENSITY_UNIT_EASY
ADJUST_DENSITY_EASY
EMPTY_HEIGHT_EASY
FULL_HEIGHT_EASY
PROCESS_DENSITY_EASYMEASURED_LEVEL_EASY
FULL_CALIBRATION_EASY
EMPTY_CALIBRATION_EASY
FULL_PRESSURE_EASY
EMPTY_PRESSURE_EASY

Write MODE_BLK

Reset codes

x
x
x
x

OOS
OOS
OOS
OOS

7864, 333

7864, 333

x
x
x
x
x

OOS
OOS
OOS
OOS
OOS

7864, 333
7864, 333
7864, 333
7864, 333

x
x
x
x

OOS
OOS
OOS
OOS

7864, 333
7864, 333
7864, 333
7864, 333

Service Transducer Block
Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage ReadClass

DEVICE_DIALOG
SW_LOCKSTATUS_LOCKINGCONFIGURATION_COUNTERELECTRONICS_TEMPERATURE
ELECTRONICS_TEMP_LOW_LIMITELECTRONICS_TEMP_HIGH_LIMITPMAX_PROC_CONN
SENSOR_MEAS_TYPE
SENSOR_MIN_ABSOLUTE_LIMITSENSOR_MAX_ABSOLUTE_LIMITSENSOR_TEMP_LOW_LIMITSENSOR_TEMP_HIGH_LIMITSENSOR_HARDWARE_REV
COUNTER P_MAX
MAX_MEASURED_PRESSURE
COUNTER_PMIN
MIN_MEASURED_PRESSURE
COUNTER_TMAX
MAX_MEASURED_TEMP
COUNTER_TMIN
MIN_MEASURED_TEMP
ELECTRONIC_OVER_TEMP_COUNTERELECTRONIC_OVER_TEMPERATURE
ELECTRONIC_UNDER_TEMP_COUNTERELECTRONIC_UNDER_TEMPERATURERESET_PEAK_HOLD
PRESSURE
CORRECTED_PRESSURE
MEASURED_VALUE_TREND
MAX_TURNDOWNSENSOR_CHANGESPRESSURE_PEAK_HOLD_STEP
TEMP_PEAK_HOLD_STEP
ACCELERATION_OF_GRAVITYHISTOROM_SAVING_CYCLE_TIMEHISTOROM_AVAIBLEDOWNLOAD_SELECTIONHISTOROM_CONTROLPRESSURE_UNIT
TEMPERATURE_UNIT
INPUT_PRESSURE_INVERSION

Device dialog
Insert PIN no.
Status locking
Config recorder
Pcb temperature
Allowed min. temp
Allowed max temp
Pmax PROC. CONN.
Sensor meas. type
Pmin sensor. damage
Pmax sensor. damage
Tmin sensor
Tmax sensor
Sens. H/ware rev.
Counter: P> Pmax
Max. meas. press.
Counter: P < Pmin
Min. meas. press.
Counter: T > Tmax
Max. meas. temp.
Counter: T < Tmin
Min. meas. temp.
Pcb count: T > Tmax
Pcb max. temp
Pcb count: T < Tmin
PCB min. temp.
Reset peakhold
Pressure
Corrected press.
Meas. val. trend
Max. turndown
Sensor changes
P. peakhold stepT. peakhold stepAcc. of gravity
Hist. saving cycl
Historom avail.
Download select.
Historom control
Cal. unit
Temp. eng. unit
Inp.press invers

11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
47
48
49
50
51
52
53

Unsigned8
Unsigned16
Unsigned16
Unsigned16
Float
Float
Float
Float
Unsigned16
Float
Float
Float
Float
Unsigned8
DS-65
DS-65
Unsigned16
Float
Unsigned16
Float
Unsigned16
Float
Unsigned16
Float
Unsigned16
Float
Unsigned8
DS-65
Float
Unsigned8
Float
Unsigned16
Float
Float
Float
Unsigned8
Unsigned8
Unsigned8
Unsigned8
Unsigned16
Unsigned16
Unsigned8

1
2
2
2
4
4
4
4
2
4
4
4
4
1
5
5
2
4
2
4
2
4
2
4
2
4
1
5
4
1
4
2
4
4
4
1
1
1
1
2
2
1

D
S
D
S
D
S
S
S
S
S
S
S
S
S
D
D
D
D
D
D
D
D
D
D
D
D
D
D
D
D
S
S
S
S
S
S
S
D
D
S
S
S

1)

x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x
x

Write MODE_BLK

Reset codesx

AUTO, OOS, MAN

7864, 333

x

AUTO, OOS, MAN

x

AUTO, OOS, MAN

x1)
x)
x)
x)
x)
x)

OOS

x
x

AUTO, OOS, MAN

x)

OOS

Can be written with service code

Endress+Hauser

43

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Display Transducer Block
Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage Read
Class

Writ
e

BLK_MODE

Reset codes

DEVICE_DIALOGDISPLAY_MAINLINE_CONTENT

Device dialog
Main line cont.

10
11

Unsigned8
Unsigned8

1
1

D
S

x
x

x

7864

DISPLAY_MAINLINE_FORMAT

Main data format

12

Unsigned8

1

S

x

x

DISPLAY_ALTERNATING_VALUES

Alternate data

13

Unsigned8

1

S

x

x

DISPLAY_CONTRAST

Display contrast

14

Unsigned8

1

S

x

x

DISPLAY_LANGUAGE

Language

15

Unsigned8

1

S

x

x

AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN

SIL_DIGITS_TEST_STRING

Digits set

16

Visible String 16

D

x

7864
7864
7864
7864

Diagnostic Transducer Block
Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage Read
Class

Writ
e

BLK_MODE

Reset codes

DEVICE_DIALOG
SW_LOCK

Device dialog
Insert PIN no.

10
11

Unsigned8
Unsigned16

1
2

D
S

x
x

x

AUTO, OOS,
MAN

7864, 333

STATUS_LOCKINGSIMULATION_MODE
SCALE_OUT_UNITS_INDEXSIMULATED_VALUE

Status locking
Simulation
Units index
Simulated value

12
13
14
15

Unsigned16
Unsigned8
Unsigned16
Float

2
1
2
4

D
D
S
D

x
x
x
x

x

OOS

x

SIMULATION_ERROR_NUMBER

Sim. error no.

16

Unsigned16

2

D

x

AUTO, OOS,MAN
AUTO, OOS,MAN

ALARM_STATUS
ALARM_STATUS_WITH_CATEGORYLAST_DIAGNOSTIC_CODE
LAST_DIAGNOSTIC_CODE_WITH_CATE
GORY
ACKNOWLEDGE_ALARM_MODE

Alarm status info
Last diag. code info

17
18
19
20

Unsigned16
Unsigned16
Unsigned16
Unsigned16

2
2
2
2

D
D
D
D

x
x
x
x

Ack. alarm mode

21

Unsigned8

1

S

x

x

ACKNOWLEDGE_ALARM

Ack. alarm

22

Unsigned8

1

D

x

x

RESET_ALL_ALARMS

Reset all alarms

23

Unsigned8

1

D

x

x

ERROR_NUMBER

Error no.

24

Unsigned16

2

D

x

x

SELECT_ALARM_TYPE

Select alarm type25

Unsigned8

1

D

x

x

ALARM_DELAY

Alarm delay

26

Float

4

S

x

x

ALARM_DISPLAY_TIME

Alarm displ. time

27

Float

4

S

x

x

PRESSURE_UNIT
PMIN_ALARM_WINDOW

Cal. unit
Pmin alarm window

28
29

Unsigned16
Float

2
4

S
S

x
x

x

PMAX_ALARM_WINDOW

Pmax alarm window

30

Float

4

S

x

x

TEMPERATURE_UNIT
TMIN_ALARM_WINDOW

Temp. eng. unit
Tmin. alarm window

31
32

Unsigned16
Float

2
4

S
S

x
x

x

TMAX_ALARM_WINDOW

Tmax. alarm window

33

Float

4

S

x

x

ENTER_RESET_CODE

Enter reset code

34

Unsigned16

2

D

x

x

OPERATING_HOURSSTATUS_HISTORYHIGHEST_CATEGORY
FF912_CONFIG_AREA

Operating hours
Alarm history
FF912ConfigArea

35
36
37
38

Unsigned32
Visible String
Unsigned8
DS271

4
18
1
30

D
D
D
S

x
x
x
x

x

FF912_STATUS_SELECT1

Status Select Event 115

39

Enumerated

1

S

x

x

FF912_STATUS_SELECT2

Status Select Event 120

40

Enumerated

1

S

x

x

FF912_STATUS_SELECT3

Status Select Event 715

41

Enumerated

1

S

x

x

44

x

AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN

AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,
MAN

7864

7864
7864
7864, 333
7864
7864
7864, 333
7864
7864

7864
7864
7864
7864

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Operation

Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage Read
Class

Writ
e

BLK_MODE

Reset codes

FF912_STATUS_SELECT4

Status Select Event 717

42

Enumerated

1

S

x

x

7864

FF912_STATUS_SELECT5

Status Select Event 718

43

Enumerated

1

S

x

x

FF912_STATUS_SELECT6

Status Select Event 720

44

Enumerated

1

S

x

x

FF912_STATUS_SELECT7

Status Select Event 726

45

Enumerated

1

S

x

x

FF912_STATUS_SELECT8

Status Select Event 727

46

Enumerated

1

S

x

x

FF912_STATUS_SELECT9

Status Select Event 730

47

Enumerated

1

S

x

x

FF912_STATUS_SELECT10

Status Select Event 731

48

Enumerated

1

S

x

x

FF912_STATUS_SELECT11

Status Select Event 732

49

Enumerated

1

S

x

x

FF912_STATUS_SELECT12

Status Select Event 733

50

Enumerated

1

S

x

x

FF912_STATUS_SELECT13

Status Select Event 740

51

Enumerated

1

S

x

x

AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,
MAN

7864
7864
7864
7864
7864
7864
7864
7864
7864

Analog Input Blocks
Parameter name,option "Symbolic name"Parameter name,option "Label"

Index

Data type

Size(Byte)

Storage ReadClass

Write BLK_MODE

FSAFE_TYPE
FSAFE_VALUE

Fsafe_Type
Fsafe_Value

37
38

Unsigned8
Float

1
4

S
S

x
x

x
x

HIHI_ALM_OUT_D

39

DS66

2

D

x

x

40

DS66

2

D

x

x

41

DS66

2

D

x

x

42

DS66

2

D

x

x

ALARM_MODE

High high alarm output
discrete
High alarm output
discrete
Low alarm output
discrete
Low low alarm output
discrete
Select alarm mode

43

Unsigned8

1

S

x

x

ALARM_OUT_D

Alarm output discrete

44

DS66

2

D

x

x

BLOCK_ERR_DESC_1

Block error description

45

Unsigned32

4

D

x

HI_ALM_OUT_D
LO_ALM_OUT_D
LOLO_ALM_OUT_D

6.3.9

Reset codes

OOS, MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,MAN
AUTO, OOS,
MAN

Methods

The FOUNDATION Fieldbus Specification includes the use of methods to make device
operation easier. A method is a sequence of interactive steps to be carried out in the specified
order so as to configure certain device functions.
The following methods are available for Cerabar S:• Restart (Resource Block)• Troubleshooting information, Config. Error No, Alarm Table (Diagnostic Block)• HistoROM (Service Block)• Sensor Trim (TRD Block)For further information on accessing methods, see the description of the FF configuration
program used.

Endress+Hauser

45

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.4

Local operation – onsite display connected

If the onsite display is connected, the three operating keys are used to navigate through theoperating menu,  ä 33, Section 6.2.3 "Function of the operating elements – onsite display
connected".

6.4.1

Menu structure

The menu is split into four levels. The three upper levels are used to navigate while you use
the lowest level to enter numerical values, and select and save options.
The structure of the MEASURING MENU depends on the measuring mode selected, e.g. if"Pressure" is selected as the measuring mode, only the functions needed for this measuringmode are displayed.

Measured value

GROUP SELECTIONLANGUAGE

MEASURING MODE

OPERATING MENUQUICK SETUP➁➀

SETTINGSDISPLAYPOSITION ADJUSTMENT

BASIC SETUP

POS. ZERO ADJUST➂➃EXTENDED SETUP

POS. INPUT VALUECALIB. OFFSET
P01-xxxxxxxx-19-xx-xx-en-097

Fig. 24:
1
2
3
4

Menu structure
1st selection level
2nd selection level
Function groups
Parameters

The MEASURING MODE parameter is only displayed via the onsite display on the 1st
selection level. In FieldCare, the LANGUAGE parameter is displayed in the DISPLAY groupand the parameters for configuring the measuring mode are displayed in the Measuring
Mode menu.

46

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.4.2

Operation

Selecting an option

Example: Selecting the "Pressure" measuring mode.
Onsite display

Operation"Level" has been selected as the measuring mode. The
option currently active is indicated by a 3 in front of themenu text.

MEASURINGMODE_Flow

Use "+" or "–" to select "Pressure" as the operating mode.

MEASURINGMODE_Press

1. Press "E" to confirm your choice. The option
currently active is indicated by a 3 in front of themenu text. (The "Pressure" measuring mode isselected.)
2. Go to the next menu item with "E"
MEASURINGMODE_Press-1

6.4.3

Editing a value

Example: changing the DAMPING VALUE function from 2.0 s to 30.0 s. See also  ä 33,
Section 6.2.3 "Function of the operating elements – onsite display connected".
Onsite display

Operation
The onsite display indicates the parameter to bemodified. The value highlighted in black can be
modified. The unit "s" is prespecified and cannot bechanged.
P01-xxxxxxxx-19-xx-xx-en-023

1. Press "+" or "–" to enter the editing mode.
2. The first digit is highlighted in black.

P01-xxxxxxxx-19-xx-xx-en-027

1. Use the "+" key to change the digit "2" to "3".
2. Press the "E" key to confirm "3". The cursor goes tothe next position (highlighted in black).

P01-xxxxxxxx-19-xx-xx-en-028

The decimal point is highlighted in black. This means
you can now edit this digit.

P01-xxxxxxxx-19-xx-xx-en-029

Endress+Hauser

47

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Onsite display

Operation
1.

Press "+" or "–" until "0" is displayed.

2. Press the "E" key to confirm "0".
The cursor goes to the next position.  is displayed
and highlighted in black. See next graphic.
P01-xxxxxxxx-19-xx-xx-en-030

Press "E" to save the new value and exit the editing
mode.  See next graphic.

P01-xxxxxxxx-19-xx-xx-en-031

The new value for the damping is 30.0 s.
– Go to the next parameter with "E"– Press "+" or "–" to go back to the editing mode.

P01-xxxxxxxx-19-xx-xx-en-032

6.4.4

Accepting pressure present at device as value

Example: performing position adjustment.
Onsite display

Operation
The bottom line on the onsite display displays thepressure present, 3.9 mbar in this example.

P01-xxxxxxxx-19-xx-xx-en-158

Use "+" or "–" to switch to the "Confirm" option. The
active option is highlighted in black.

P01-xxxxxxxx-19-xx-xx-en-159

Using the "E" key, assign the value (3.9 mbar) to the
POS. ZERO ADJUST parameter. The device confirms theadjustment and goes back to the parameter, here POS.
ZERO ADJUST (see next graphic).
P01-xxxxxxxx-19-xx-xx-en-037

Go to the next parameter with "E"

P01-xxxxxxxx-19-xx-xx-en-160

48

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.5

Operation

FieldCare

FieldCare is Endress+Hauser's plant asset management tool based on FDT technology.
You can use FieldCare to configure all Endress+Hauser devices as well as third-party devices
which support the FDT standard. Hardware and software requirements you can find on theinternet: www.endress.com  select your country  Search: FieldCare  FieldCare ®
Technical Data.
FieldCare supports the following functions:• Configuration of transmitters in offline and online mode
• Loading and saving device data (upload/download)• HistoROM®/M-DAT analysis• Documentation of the measuring point
Connection options:• Service interface with Commubox FXA291 and ToF adapter FXA291 (USB).
• In the "Level Standard" measuring mode, the configuration data that were loaded with FDT
upload cannot be written again (FDT download). These data are only used to document themeasuring point.
• For further information, see  www.endress.com

6.6

HistoROM®/M-DAT (optional)

NOTICE

Device could be destroyed!
Detach HistoROM®/M-DAT from the electronic insert or attach it to the insert in adeenergized state only.
HistoROM®/M-DAT is a memory module, which is attached to the electronic insert andfulfills the following functions:• Back-up copy of configuration data
• Copying configuration data from one transmitter to another transmitter• Cyclic recording of pressure and sensor-temperature measured values• Recording diverse events, such as alarms, configuration changes, counters for measuring
range undershoot and overshoot for pressure and temperature, overshooting andundershooting user limits for pressure and temperature, etc.
• The HistoROM®/M-DAT can be retrofitted at any time (Order No.: 52027785).
• To analyze and evaluate the data and events saved in the HistoROM®/M-DAT, you require
the Endress+Hauser FieldCare operating program. A CD with the operating program anddocumentation is provided for devices that were ordered with the "HistoROM/M-DAT"
option.
 ä 49, Section 6.5 "FieldCare". It is also possible to copy configuration data from one
transmitter to another transmitter with an FF configuration program.
• The HistoROM data and the data in the device are analyzed once a HistoROM ®/M-DAT is
attached to the electronic insert and power is reestablished to the device. During theanalysis, the messages "W702, HistoROM data not consistent" and "W706, Configurationin HistoROM and device not identical" can occur. For remedial measures,  ä 81,
Section 9.2 "Diagnostic information on local display."

Endress+Hauser

49

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.6.1

Copying configuration data➁

Sim.

Operation

on

off

1

2

➀

0%
Zero
Display

HW

on
2

off

PC

1

1
2

250002273-–

TM

FOUNDATION

HW-Version:
SW-Version:

Histo ROM

Simulation

Sensor

P01-xxxxxxxx-19-xx-xx-xx-123

Electronic insert with optional HistoROM®/M-DAT memory module
1
2

Optional HistoROM®/M-DAT
To copy configuration data from the HistoROM/®M-DAT to a device or from a device to a HistoROM®/M-DAT module,
operation must be unlocked (DIP switch 1, position "Off", parameter SWLOCK/INSERT PIN No = 100). See also  ä 52,
Section 6.7 "Locking/unlocking operation".

Local operation via onsite display (optional) or remote operation
Copying configuration data from a device to a HistoROM®/M-DAT:Operation must be unlocked.

50

1.

Disconnect the device from the supply voltage.

2.

Remove the protection cap, attach the HistoROM®/M-DAT to the electronic insert.

3.

Reestablish supply voltage to the device.

4.

The option selected for the DOWNLOAD SELECT. parameter (OPERATION menu) does
not affect uploading from the device to the HistoROM.

5.

Operation via an FF configuration program: using the DAT_HANDLING/ HistoROM
CONTROL parameter in the Service Transducer Block, select the "Device  HistoROM"
option for the data transfer direction.
Operation via FieldCare: using the HistoROM CONTROL parameter, select the "Device 
HistoROM" option for the data transfer direction. (Menu path: OPERATING MENU 
OPERATION)
Using the DOWNLOAD SELECT. parameter (OPERATION menu), select whichparameters should be overwritten.
The following parameters are overwritten depending on the option selected:– Configuration copy:
All the parameters apart from the TRANSMITTER SERIAL NO., DEVICEDESIGNATION and the parameters of the POSITION ADJUSTMENT and PROCESSCONNECTION group.
– Device replacement:
All the parameters apart from the TRANSMITTER SERIAL NO., DEVICEDESIGNATION and the parameters of the POSITION ADJUSTMENT and PROCESSCONNECTION group.
– Electronics replace:
All parameters apart from the parameters of the POSITION ADJUSTMENT group
Factory setting: Configuration copy

6.

Using the HistoROM CONTROL parameter, select the "Device HistoROM" option forthe data transfer direction.

7.

Wait approx. 40 seconds. Configuration data are uploaded from the device to theHistoROM®/M-DAT. The device is not restarted.

8.

Disconnect the device from the supply voltage again.

9.

Detach the memory module.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Operation

10. Reestablish supply voltage to the device.
Copying configuration data from a HistoROM®/M-DAT to a device:
Operation must be unlocked.

Endress+Hauser

1.

Disconnect the device from the supply voltage.

2.

Attach the HistoROM®/M-DAT to the electronic insert. Configuration data fromanother device are stored in the HistoROM®/M-DAT.

3.

Reestablish supply voltage to the device.

4.

Operation via an FF configuration program: using the DAT_HANDLING/ HistoROM
CONTROL parameter in the Service Transducer Block, select the "HistoROM  Device"
option for the data transfer direction.
Operation via FieldCare: using the HistoROM CONTROL parameter, select the"HistoROM  Device" option for the data transfer direction (menu path: OPERATINGMENU  OPERATION).
Using the DOWNLOAD SELECT. parameter (OPERATION menu), select whichparameters should be overwritten.
The following parameters are overwritten depending on the option selected:– Configuration copy (factory setting)
All the parameters apart from the DEVICE SERIAL No., DEVICE DESIGN, PD-TAG,
DESCRIPTION, DEVICE ID, DEVICE ADDRESS and the parameters in the POSITION
ADJUSTMENT, PROCESS CONNECTION, SENSOR TRIM and SENSOR DATA group.
– Device replacement
All the parameters apart from the DEVICE SERIAL No., DEVICE ID, DEVICE DESIGN
and the parameters in the POSITION ADJUSTMENT, PROCESS CONNECTION,SENSOR TRIM and SENSOR DATA group.
– Electronics replace
All parameters apart from the parameters of the SENSOR DATA group.
Factory setting: Configuration copy

5.

Using the HistoROM CONTROL parameter (OPERATION menu), select the "HistoROM
Device" option for the data transfer direction.

6.

Wait approx. 40 seconds. Configuration data are loaded from the HistoROM®/M-DAT
to the device. The device is restarted.

7.

Before removing the HistoROM®/M-DAT again from the electronic insert, disconnect
the device from the supply voltage.

51

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.7

Locking/unlocking operation

Once you have entered all the parameters, you can lock your entries against unauthorized
and undesired access.
The operation can be locked/unlocked in the following ways:• Via a DIP switch on the electronic insert, locally on the display.
• Via communication e.g. FieldCare.
The -symbol on the onsite display indicates that operation is locked. Parameters whichrefer to how the display appears, e.g. LANGUAGE and DISPLAY CONTRAST, can still be
altered.
• If operation is locked by means of the DIP switch, you can only unlock operation again bymeans of the DIP switch. If operation is locked by means of remote operation e.g.
FieldCare, you can only unlock operation again by means of remote operation.
The table provides an overview of the locking functions:
View/readparameter

Modify/write via1)

DIP switch

Yes

Remote operation

Yes

Locking via

DIP switch

Remote operation

No

Yes

No

No

No

Yes

Parameters which refer to how the display appears, e.g. LANGUAGE and DISPLAY CONTRAST, can still be
altered.

Locking/unlocking operation locally via DIP switch

➂
on

0%
Zero

on

Display

Sensor

HW

on

1

2

off

1

2

2

off

PC

1

off

–

+

1
2

250002273-–

TM

FOUNDATION

HW-Version:
SW-Version:

Histo ROM➁

Simulation➀

Simulation

6.7.1

Simulation1)

Unlocking via

E

P01-xxxxxxxx-19-xx-xx-xx-177

Fig. 25:
1
2
3"Hardware locking" DIP switch position on the electronic insert
If necessary, remove onsite display (optional)
DIP switch is set to "On": operation is locked.
DIP switch is set to "Off": operation is unlocked (operation possible)

6.7.2

Locking/unlocking operation via remote operation
Description

Locking operation

1. Operation via FF configuration program: select SWLOCK parameter in the
Resource Block.
Operation via FieldCare: select INSERT PIN No. parameter, menu path:
OPERATING MENU OPERATION INSERT PIN No.
2. To lock operation, enter a number for this parameter between 0...9999 that is 100.

Unlocking operation

1. Operation via FF configuration program: select SWLOCK parameter in the
Resource Block.
Operation via FieldCare: select INSERT PIN No. parameter.
2. To unlock operation, enter "100" for the parameter.

52

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.8

Operation

Simulation

The function of the Analog Input Block, such as input and output scaling, can be simulated
as follows:1.

Set the "Simulation" DIP switch on the electronic insert to "On".

2.

In the Analog Input Block, select the "Active" option by means of the SIMULATION
parameter, ENABLE_DISABLE element.

3.

Set the Analog Input Block to the AUTO block mode.

4.

Enter the value and status for the SIMULATION_VALUE and SIMULATION_STATUS
elements. During the simulation, the output value and status of the Pressure
Transducer Block are replaced by the simulated value and status. The OUT parametershows the result.

5.

End simulation (SIMULATION parameter, ENABLE_DISABLE element, "Disabled"
option).

You can check your adjustment for the transmitter by means of the SIMULATION_MODE
and SIMULATION_VALUE parameters in the Diagnostic Transducer Block. See Operating
Instructions BA00303 "Description of Device Functions Cerabar S/Deltabar S/Deltapilot S",
SIMULATION_MODE and SIMULATION_VALUE parameter descriptions.

6.9

Factory setting (reset)• Total reset: Press zero-key for at least 12 seconds. The LED on the electronic insert lights
up briefly if a reset is being carried out.
• By entering a certain code, you can completely, or partially, reset the entries for theparameters to the factory settings. (For the factory settings, see Operating Instructions
BA00303P "Cerabar S/Deltabar S/Deltapilot S, Description of Device Functions").
Enter the code by means of the ENTER RESET CODE parameter (OPERATION menu).
There are various reset codes for the device. The following table illustrates whichparameters are reset by the particular reset codes. Operation has to be unlocked to be able
to perform a reset ( ä 52, Section 6.7 "Locking/unlocking operation").
• Any customer-specific configuration carried out by the factory is not affected by a reset(customer-specific configuration remains). If, after a reset, you wish the parameters to bereset to the factory settings, please contact Endress+Hauser Service.
• The OUT parameter may have to be rescaled after resetting with code 7864.  ä 67,
Section 7.8 "Scaling the OUT parameter".

6.9.1

Performing reset via an FF configuration program

If operating via an FF configuration program, enter the code by means of the
RESET_INPUT_VALUE/ENTER RESET CODE parameter in the Diagnostic Transducer Block.
The index tables  ä 41 ff. indicate which parameters are reset by the particular reset code.
The RESET FF parameter gives you the option of deleting links between function blocks andresetting FF parameters to default values and manufacturer-specific parameters to thefactory setting. See also Operating Instructions BA00303P, RESTART parameter
description.

Endress+Hauser

53

Operation

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

6.9.2

Performing reset via the FieldCare operating program

If operating via FieldCare, enter the code via the ENTER RESET CODE parameter (menu path:
OPERATING MENU OPERATION).
The following table illustrates which parameters are reset by the particular reset codes.
Reset code

Description and effect 1)

7864

Total reset– This reset resets the following parameters:– POSITION ADJUSTMENT function group– BASIC SETUP function group– EXTENDED SETUP function group– LINEARIZATION function group (an existing linearization table is deleted)– OUTPUT group– INFO function group, TAG_DESC parameter– MESSAGES function group– All configurable messages ("Error" type) are set to "Warning".
 ä 81, Section 9.2 "Diagnostic information on local display" and  ä 95, Section 9.6"Response of outputs to errors".
– USER LIMITS function group– Any simulation which may be running is ended.
– The device is restarted.

333

User reset– This reset resets the following parameters:– POSITION ADJUSTMENT function group– BASIC SETUP function group, apart from customer-specific units– EXTENDED SETUP function group– OUTPUT group– Any simulation which may be running is ended.
– The device is restarted.

2710

Reset Level measuring mode
– Depending on the settings for the LEVEL MODE, LIN MEASURAND, LINd MEASURAND or
COMB. MEASURAND parameters, the parameters needed for this measuring task will bereset.
– Any simulation which may be running is ended.
– The device is restarted.
Example LEVEL MODE = linear and LIN. MEASURAND = level• HEIGHT UNIT = m• CALIBRATION MODE = wet• EMPTY CALIB. = 0• FULL CALIB. = sensor end value converted to mH2O, e.g. for a 500 mbar (7.5 psi) sensor :50.99 mH2O

54

2509

Sensor adaption reset– This reset resets the upper and lower sensor calibration limit and the value for position
adjustment.
– POSITION ADJUSTMENT function group– PRESSURE_1_LOWER_CAL/LO_TRIM_MEASURED and
PRESSURE_1_HIGHER_TRIM_MEASURED/HI_TRIM_MEASURED parameters
These parameters are not available by means of the FieldCare operating program.
– Any simulation which may be running is ended.
– The device is restarted.

1846

Display reset– This reset resets all parameters which have to do with how the display appears (DISPLAY
group).
– Any simulation which may be running is ended.
– The device is restarted.

8888

HistoROM reset
The measured value and event buffers are deleted. During the reset, the HistoROM has to be
attached to the electronic insert.

62

PowerUp reset (warm start)– This reset resets all the parameters in the RAM. Data are read back anew from the
EEPROM (processor is initialized again).
– Any simulation which may be running is ended.
– The device is restarted.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

1)

Endress+Hauser

Operation

The table uses the group and parameter names as they appear in FieldCare.  ä 41, Section 6.3.8 "Index
tables of Endress+Hauser parameters" for the assignment of the FieldCare parameter names and the FF
configuration program.

55

Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7

Commissioning

The device is configured for the Pressure measuring mode as standard. The measuring range
and the unit in which the measured value is transmitted correspond to the specifications on
the nameplate.
!

WARNING

Exceeding the maximum allowable working pressure!
Risk of injury due to bursting of parts! Warning messages are generated if pressure is too
high.
‣ If a pressure greater than the maximum permitted pressure is present at the device, themessages "E115 Sensor overpressure" and "E727 Sensor pressure error - overrange" areoutput in succession! Use the device only within the sensor range limits
NOTICE

Shortfall of the allowable working pressure!
Output of messages if pressure is too low.
‣ If a pressure smaller than the minimum permitted pressure is present at the device, themessages "E120 Sensor low pressure" and "E727 Sensor pressure error - overrange" areoutput in succession! Use the device only within the sensor range limits

7.1

Configuring messages• The messages E727, E115 and E120 are "Error"-type messages and can be configured as a"Warning" or an "Alarm". The factory setting for these messages is "Warning". This setting
prevents the BAD status from being transmitted in applications (e.g. cascademeasurement) where the user is aware of the risk of the sensor range being overshot.
• We recommend setting messages E727, E115 and E120 to "Alarm" in the following
instances:– It is not necessary to violate the sensor range for the measuring application.
– A position adjustment must be carried out that has to correct a large measured error asa result of the orientation of the device.

7.2

Function check

Carry out a post-installation and a post-connection check as per the checklist before
commissioning the device.
• "Post-installation check" checklist see Section 4.5• "Post-connection check" checklist see Section 5.4

7.3

Commissioning via an FF configuration program• The device is configured for the Pressure measuring mode as standard. The measuring
range and the unit in which the measured value is transmitted, as well as the digital output
value of the Analog Input Block OUT, correspond to the data on the nameplate. Following
a reset with code 7864, the OUT parameter may have to be rescaled ( ä 67, Section 7.8"Scaling the OUT parameter").
• The standard order configuration is illustrated  ä 36, Section 6.3.6 "Cerabar S blockmodel".

56

1.

Switch on the device.

2.

Note the DEVICE_ID.  ä 36, Section 6.3.5 "Device identification and addressing" and ä 10, Section 3.2 "Device designation" for the device serial number.

3.

Open the configuration program.

4.

Load Cff and device description files into the host system or the configuration program.
Make sure you are using the right system files.
Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

5.

Commissioning

Identify the device using the DEVICE_ID (see Point 2). Assign the desired tag nameto the device by means of the PD_TAG parameter.

Configuring the Resource Block
1.

Open the Resource Block.

2.

If necessary, disable the lock for device operation.  ä 52, Section 6.7 "Locking/unlocking operation". Operating is unlocked as standard.

3.

If necessary, change the block name. Factory setting: RS_452B481007-xxxxxxxxxxx

4.

If necessary, assign a description to the block by means of the TAG_DESC parameter.

5.

If necessary, change other parameters as per the requirements.

Configuring the Transducer Blocks
Cerabar S has the following Transducer Blocks:• Pressure Transducer Block• Service Transducer Block• Display Transducer Block• Diagnostic Transducer Block
The explanation that follows is an example for the Pressure Transducer Block.
1.

If necessary, change the block name. Factory setting: RS_452B481007-xxxxxxxxxxx

2.

Set the block mode to OOS using the MODE_BLK parameter, TARGET element.

3.

Configure the device in accordance with the measuring task. See also these
Operating Instructions Section 7.4 to Section 7.8.

4.

Set the block mode to Auto using the MODE_BLK parameter, TARGET element.

The block mode must be set to "Auto" for the Pressure and Service Transducer Block for themeasuring device to function correctly.
Configuring the Analog Input Blocks
Cerabar S has 2 Analog Input Blocks that can be assigned as required to the various process
variables.

Endress+Hauser1.

If necessary, change the block name. Factory setting: RS_452B481007-xxxxxxxxxxx

2.

Set the block mode to OOS using the MODE_BLK parameter, TARGET element.

3.

Use the CHANNEL parameter to select the process variable which should be used as the
input value for the Analog Input Block. The following settings are possible:– CHANNEL = 1: Primary value, a pressure or level value depending on the measuringmode selected– CHANNEL = 2: Secondary value, here the sensor temperature
Factory setting:– Analog Input Block 1: CHANNEL = 1: Primary Value (pressure measured value)– Analog Input Block 2: CHANNEL = 2: Secondary Value (sensor temperature)

4.

Use the XD_SCALE parameter to select the desired unit and the block input range forthe process variable.  ä 67, Section 7.8 "Scaling the OUT parameter".
Make sure that the unit selected suits the process variable selected. If the process
variable does not suit the unit, the BLOCK_ERROR parameter reports "Block
Configuration Error" and the block mode cannot be set to "Auto".

5.

Use the L_TYPE parameter to select the type of linearization for the input variable(factory setting: Direct).
Make sure that the settings for the XD_SCALE and OUT_SCALE parameters are thesame for the "Direct" linearization type. If the process values and units do not match, the
BLOCK_ERROR parameter reports "Block Configuration Error" and the block modecannot be set to "Auto".

6.

Enter the alarm and critical alarm messages by means of the HI_HI_LIM, HI_LIM,
LO_LIM and LO_LO_LIM parameters. The limit values entered have to be within thevalue range specified for the OUT_SCALE parameter.
57

Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.

Specify the alarm priorities by means of the HI_HI_PRI, HI_PRI, LO_LO_PRI and LO_PRI
parameters. Reporting to the field host system only takes place with alarms with apriority greater than 2.

8.

Set the block mode to Auto using the MODE_BLK parameter, TARGET element. For thispurpose, the Resource Block must also be set to the "Auto" block mode.

Additional configuration1.

Depending on the control or automation task, configure additional function blocks andoutput blocks. See also Operating Instructions BA00303P "Description of Device
Functions Cerabar S/Deltabar S/Deltapilot S".

2.

Link the function blocks and output blocks.

3.

After specifying the active LAS, download all the data and parameters to the fielddevice.

7.4

Selecting the language and measuring mode

7.4.1

Local operation

The MEASURING MODE parameter is on the 1st selection level.
Section 6.4.1 "Menu structure".
The following measuring modes are available:• Pressure• Level58

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.4.2

Commissioning

Selecting the language and measuring mode by means of theFieldCare operating program

Selecting the measuring mode
The parameters for setting the measuring mode are displayed in the FieldCare "Measuringmode" menu:!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣ If the measuring mode is changed, the span setting (URV) must be verified in the"CALIBRATION"  "BASIC SETUP" operating menu and, if necessary, reconfigured!

Betriebsart-de

Fig. 26:"Measuring mode" menu

The following measuring mode settings are available:
Primary Value TypeLinearization

Level selection

Pressure

None-

Level, mass, volume

None

Level Easy Pressure

Level, mass, volume

None

Level Easy Height

Level, mass, volume, tank content in %None

Level Standard

Level, mass, volume, tank content in %Level linearized

Level Standard

Level, mass, volume, tank content in %Level combined

Level Standard

Selecting the language
Select the menu language for FieldCare using the "Language Button" in the configuration
window. Select the menu language for the FieldCare frame by means of the "Extra" menu "Options" "Display"  "Language".
The following languages are available:• Deutsch• English• Français• Español• Chinese• JapaneseEndress+Hauser

59Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.5

Position adjustment

Due to the orientation of the device, there may be a shift in the measured value, i.e. when
the container is empty or partly filled, the measured value parameter does not display zero.
There are two options to choose from when performing position adjustment.
• Onsite display menu path:
GROUP SELECTION  OPERATING MENU SETTINGS POSITION ADJUSTMENT• FieldCare menu path:
OPERATING MENU SETTINGS POSITION ADJUSTMENT

7.5.1

Performing position adjustment via the onsite display or
FieldCare

The parameters listed in the following table can be found in the POSITION ADJUSTMENT
group (menu path: OPERATING MENU SETTINGS POSITION ADJUSTMENT).
Parameter name

Description

POS. ZERO ADJUST
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure need not be known.
Example:– MEASURED VALUE = 2.2 mbar (0.032 psi)
– Correct the MEASURED VALUE via the POS. ZERO ADJUST parameter with the"Confirm" option. This means that you are assigning the value 0.0 to the pressure
present.
– MEASURED VALUE (after pos. zero adjust) = 0.0 mbar
The CALIB. OFFSET parameter displays the resulting pressure difference (offset) bywhich the MEASURED VALUE was corrected.
Factory setting:
0.0

POS. INPUT VALUE
Input

Position adjustment – the pressure difference between zero (set point) and themeasured pressure need not be known. To correct the pressure difference, youneed a reference measurement value (e. g. from a reference device).
Example:– MEASURED VALUE = 0.5 mbar (0.0073 psi)
– For the POS. INPUT VALUE parameter, specify the desired set point for the
MEASURED VALUE, e.g. 2.0 mbar (0.029 psi).
(The following applies: MEASURED VALUEnew = POS. INPUT VALUE)– MEASURED VALUE (after entry for POS. INPUT VALUE) = 2.0 mbar (0.029 psi)
– The CALIB. OFFSET parameter displays the resulting pressure difference (offset)
by which the MEASURED VALUE was corrected.
The following applies: CALIB. OFFSET = MEASURED VALUEold – POS. INPUTVALUE,
here: CALIB. OFFSET= 0.5 mbar (0.0073 psi) – 2.0 mbar (0.029 psi) = – 1.5
mbar (0.022 psi)
Factory setting:
0.0

CALIB. OFFSET
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure is known. (A reference pressure is not present at the device.)
Example:– MEASURED VALUE = 2.2 mbar (0.032 psi)
– Via the CALIB. OFFSET parameter, enter the value by which the MEASUREDVALUE should be corrected. To correct the MEASURED VALUE to 0.0 mbar, youmust enter the value 2.2 here.
(The following applies: MEASURED VALUE new = MEASURED VALUEold –CALIB. OFFSET)– MEASURED VALUE (after entry for calib. offset) = 0.0 mbar
Factory setting:
0.0

60

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Commissioning

7.6

Pressure measurement

7.6.1

Information on pressure measurement• A Quick Setup menu is available for each of the measuring modes pressure and level which
guides you through the most important basic functions. With the setting in the
MEASURING MODE parameter, you specify which Quick Setup menu should be displayed.
 ä 58, Section 7.4 "Selecting the language and measuring mode".
• For a detailed description of the parameters, see Operating Instructions BA00303P"Cerabar S/Deltabar S/Deltapilot S, Description of Device Functions"– FF, Table, Pressure Transducer Block– FieldCare: Table, POSITION ADJUSTMENT– FieldCare: Table, BASIC SETUP– FieldCare: Table, EXTENDED SETUP!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣ If the measuring mode is changed, the span setting (URV) must be verified in the"CALIBRATION"  "BASIC SETUP" operating menu and, if necessary, reconfigured!

7.6.2

Quick Setup menu for the Pressure measuring mode

MEASURED VALUE

1)
GROUP SELECTION

1)LANGUAGE

1)
MEASURING MODE

2)
MEASURING MODE

1)QUICK SETUPOPERATING MENU

Pressure

Level
Absolut pressure sensors

Gauge pressure
sensors
POS. ZERO ADJUSTPOS. INPUT VALUEDAMPING VALUE

1) On-site display only
2) FieldCare only
P01-PMx7xxxx-19-xx-xx-en-078

Fig. 27:

Quick Setup menu for the "Pressure" measuring mode

Local operation

FieldCare

Measured value display
Switch from the measured value display to the
GROUP SELECTION with F .

Measured value display
Select QUICK SETUP menu.

GROUP SELECTION
Select the MEASURING MODE parameter.

Measuring Mode
Select the Primary Value Type parameter.

MEASURING MODE
Select "Pressure".

Endress+Hauser

GROUP SELECTION
Select QUICK SETUP menu.

Primary Value Type
Select "Pressure" option.

POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i.e. you assign the value 0.0 to
the pressure present.

POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i.e. you assign the value 0.0 to
the pressure present.

61

Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Local operation

FieldCare

POS. INPUT VALUE
Due to orientation of the device, there may be a shift
in the measured value. For the POS. INPUT VALUE
parameter, specify the desired set point for the
MEASURED VALUE.

POS. INPUT VALUE
Due to orientation of the device, there may be a shift
in the measured value. For the POS. INPUT VALUE
parameter, specify the desired set point for the
MEASURED VALUE.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
Value of the Analog Input Block react to a change inthe pressure.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
Value of the Analog Input Block react to a change inthe pressure.

For onsite operation, see also  ä 33, Section 6.2.3 "Function of the operating elements –onsite display connected" and  ä 46, Section 6.4 "Local operation – onsite display
connected".

62

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.7

Level measurement

7.7.1

Information on level measurement

Commissioning• A Quick Setup menu is available for each of the measuring modes Pressure and Level
which guides you through the most important basic functions. For the "Level" Quick
Setup menu,  ä 65.
• Furthermore, three level modes are available for the level measurement, namely "Leveleasy pressure", Level easy height" and "Level standard". For the "Level standard" level mode,
you can choose between the "Linear", "Pressure linearized" and "Height linearized" leveltypes. The table in the "Overview of level measurement" section that follows provides you
with an overview of the various measuring tasks.
– With regard to the "Level easy pressure" and "Level easy height" level modes, the values
entered are not tested as extensively as in the "Level standard" level mode. In the "Leveleasy pressure" and "Level easy height" level modes, the values entered for EMPTYCALIBRATION/FULL CALIBRATION, EMPTY PRESSURE/FULL PRESSURE and EMPTYHEIGHT/FULL HEIGHT have to be at least 1 % apart. The value will be rejected with awarning message if the values are too close together. Further limit values are notchecked; i.e. the values entered must be appropriate for the sensor and the measuringtask so that the measuring device can measure correctly.
– The "Level easy pressure" and "Level easy height" level modes comprise fewer parametersthan the "Level standard" mode and are used to quickly and easily configure a levelapplication.
– Customer-specific units of level, volume and mass, or a linearization table, can only beentered in the "Level standard" level mode.
• For a detailed description of the parameters and configuration examples, see Operating
Instructions BA00303P "Cerabar S/Deltabar S/Deltapilot S, Description of Device
Functions".
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣ If the measuring mode is changed, the span setting (URV) must be verified in the"CALIBRATION"  "BASIC SETUP" operating menu and, if necessary, reconfigured!

Endress+Hauser

63Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.7.2
Measuring task

LEVEL SELECTION/LEVEL MODE

Overview of level measurement
Measured variable Description
options

Comment

Measured value display

The measured variable is LEVEL SELECTION:
Level easy pressure
in direct proportion to
the measured pressure.
Calibration takes placeby entering two
pressure-level pairs.

Via OUTPUT UNIT – Calibration with reference – Incorrect entries
are possiblepressure – wetparameter: %, level,
calibration, see Operating – Customized units
volume or mass
are not possible
Instructions BA00303P.
units.
– Calibration without
reference pressure – drycalibration, see Operating
Instructions BA00303P.

The measured value
display and the LEVELBEFORE LIN. parametershow the measured value.

The measured variable is LEVEL SELECTION:
Level easy heightin direct proportion to
the measured pressure.
Calibration takes placeby entering the density
and two height-levelpairs.

Via OUTPUT UNIT – Calibration with reference – Incorrect entries
are possiblepressure – wetparameter: %, level,
calibration, see Operating – Customized units
volume or mass
are not possible
Instructions BA00303P.
units.
– Calibration without
reference pressure – drycalibration, see Operating
Instructions BA00303P.

The measured value
display and the LEVELBEFORE LIN. parametershow the measured value.

The measured variable is LEVEL SELECTION:
Level standard/in direct proportion to
the measured pressure. LEVEL MODE:
LinearVia LIN.
MEASURAND
parameter:– % (level)– Level– Volume– Mass– Calibration with reference – Incorrect entries
are rejected bypressure – wet
the device
calibration, see Operating– Customized level,
Instructions BA00303P.
volume and mass
– Calibration without
units are possiblereference pressure – drycalibration, see Operating
Instructions BA00303P.

The measured value
display and the LEVELBEFORE LIN. parametershow the measured value.

The measured value
display and the TANKCONTENT parametershow the measured value.

The measured variable is
not in direct proportion
to the measured
pressure as is the casewith tanks with a conicaloutlet, for example. A
linearization table has to
be entered for thecalibration.

LEVEL SELECTION:
level standard/
LEVEL MODE:
pressure linearized

Via LINd
MEASURAND
parameter:– Pressure + %– Pressure +volume– Pressure + mass

– Calibration with reference – Incorrect entries
pressure: semiautomatic
are rejected byentry of linearization
the devicetable, see Operating– Customized level,
Instructions BA00303P.
volume and mass
– Calibration without
units are possiblereference pressure:manual entry of
linearization table, see
Operating Instructions
BA00303P.

– Two measured
variables are requiredor– the tank shape isgiven with value pairs,e.g. height andvolume.

LEVEL SELECTION:
Level standard/
LEVEL MODE: Heightlinearized

Via COMB.
MEASURAND
parameter:– Height + volume– Height + mass
– Height + %– %-height +volume– %-height + mass
– %-height + %– Calibration with reference – Incorrect entries The measured value
display and the TANK
are rejected bypressure: wet calibration
CONTENT parameterthe device
and semiautomatic entryof linearization table, see – Customized level, show the 2nd measured
volume and mass value (volume, mass or
Operating Instructions
units are possible %).
BA00303P.
– Calibration without
The LEVEL BEFORE LIN
reference pressure: dry
parameter displays thecalibration and manual
1st measured value (%entry of linearization
height or height).
table, see Operating
Instructions BA00303P.

The 1st measured
variable %-height orheight must be in direct
proportion to themeasured pressure. The
2nd measured variable
volume, mass or % mustnot be in direct
proportion to themeasured pressure. A
linearization table must
be entered for the 2nd
measured variable. The
2nd measured variable isassigned to the 1st
measured variable bymeans of this table.

64

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.7.3

Commissioning

Quick Setup menu for the Level measuring mode

• Some parameters are only displayed if other parameters are appropriately configured. For
example, the EMPTY CALIBRATION parameter is only displayed in the following
instances:– LEVEL SELECTION "Level easy pressure" and CALIBRATION MODE "Wet"– LEVEL SELECTION "Level standard", LEVEL MODE "Linear" and CALIBRATION MODE
"Wet"
The LEVEL MODE and CALIBRATION MODE parameters are in the BASIC SETUP functiongroup.
• The following parameters are set to the following values at the factory:– LEVEL SELECTION: Level easy pressure– CALIBRATION MODE: Wet– OUTPUT UNIT or LIN. MEASURAND: %– EMPTY CALIBRATION: 0.0– FULL CALIBRATION: 100.0• The Quick Setup is suitable for easy and quick commissioning. If you want to make more
complex settings like changing the unit from "%" to "m", the calibration must be performed
via the BASIC SETUP group. See Operating Instructions BA00303P.
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
‣ If the measuring mode is changed, the span setting (URV) must be verified in the"CALIBRATION"  "BASIC SETUP" operating menu and, if necessary, reconfigured!

MEASURED VALUE

1)
GROUP SELECTION

1)LANGUAGE

1)
MEASURING MODE

2)
MEASURING MODE

1)QUICK SETUPOPERATING MENU

1)
LEVEL SELECTION

Pressure

Level
POS. ZERO ADJUST3)
EMPTY CALIB.

3)
FULL CALIB.

DAMPING VALUE

1) On-site display only
2) FieldCare only3) – LEVEL SELECTION "Level Easy Pressure"and CALIBRATION MODE "Wet"– LEVEL SELECTION = "Level Standard",
LEVEL MODE = "Linear" andCALIBRATION MODE = "Wet"
P01-FMB70xxx-19-xx-xx-en-015

Fig. 28:

Endress+Hauser

Quick Setup menu for the "Level" measuring mode"

Local operation

FieldCare

Measured value display
Switch from the measured value display to the
GROUP SELECTION with F .

Measured value display
Select QUICK SETUP menu.

GROUP SELECTION
Select the MEASURING MODE.

Measuring Mode
Select the Primary Value Type parameter.

MEASURING MODE
Select "Level".

Primary Value Type
Select "Level" option.

LEVEL SELECTION
Select the level mode. For an overview,  ä 64.

Level Selection
Select the level mode. For an overview,  ä 64.

65

Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Local operation

FieldCareGROUP SELECTION
Select QUICK SETUP menu.
POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i.e. you assign the value 0.0 to
the pressure present.

POS. ZERO ADJUSTDue to orientation of the device, there may be a shift
in the measured value. You correct the MEASUREDVALUE via the POS. ZERO ADJUST parameter withthe "Confirm" option, i.e. you assign the value 0.0 to
the pressure present.

EMPTY CALIB. 1)
Enter level value for the lower calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

EMPTY CALIBRATION 1
Enter level value for the lower calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

FULL CALIBRATION 1
Enter level value for the upper calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

FULL CALIBRATION 1
Enter level value for the upper calibration point.
For this parameter, enter a level value which is
assigned to the pressure present at the device.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
Value of the Analog Input Block react to a change inthe pressure.

DAMPING VALUEEnter damping time (time constant ). The damping
affects the speed at which all subsequent elements,
such as the onsite display, measured value and OUT
Value of the Analog Input Block react to a change inthe pressure.

1)– LEVEL SELECTION "Level easy pressure" and CALIBRATION MODE "Wet"– LEVEL SELECTION "Level standard", LEVEL MODE "Linear" and CALIBRATION MODE "Wet"

For onsite operation, see also  ä 33, Section 6.2.3 "Function of the operating elements –onsite display connected" and  ä 46, Section 6.4 "Local operation – onsite display
connected".

66

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.8

Commissioning

Scaling the OUT parameterIn the Analog Input Block, the input value or input range can be scaled in accordance withthe automation requirements.
Example:The measuring range 0 to 500 (7.5 psi) mbar should be rescaled to 0 to 100 %.
• Select XD_SCALE group.
– For EU_0, enter "0".
– For EU_100, enter "500".
– For UNITS_INDEX, enter "mbar".
• Select OUT_SCALE group.
– For EU_0, enter "0".
– For EU_100, enter "10000".
– For UNITS_INDEX, select "%" for example.
The unit selected here does not have any effect on the scaling. This unit is not displayedon the onsite display or in the operating program such as FieldCare.
• Result:
At a pressure of 350 mbar (5.25 psi), the value 70 is output to a downstream block or to
the PCS as the OUT value.
Analog Input Block
L_TYPE: Indirect

UNITS_INDEX:
mbar

XD_SCALE

EU_100500

OUT_SCALE

1PRIMARY_VALUE= 350 mbar
Output Pressure
Transducer Block

0.7

EU_0
0

0
0

0.7

1

EU_0
0

UNITS_
EU_100 INDEX:
%
70
100

OUT Value = 70
P01-xMx7xxxx-05-xx-xx-en-008!

CAUTION

Note Dependencies when setting parameters!
‣ If you have selected the "Direct" mode for the L_TYPE parameter, you cannot change thevalues and units for XD_SCALE and OUT_SCALE.
‣ The L_TYPE, XD_SCALE and OUT_SCALE parameters can only be changed in the OOS
block mode.
‣ Make sure that the output scaling of the Pressure Transducer Block SCALE_OUT matches
the input scaling of the Analog Input Block XD_SCALE.

Endress+Hauser

67Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.9

Configuring event behavior in accordance with
FOUNDATION Fieldbus Specification FF912 Field
Diagnostic Profile

The device complies with FOUNDATION Fieldbus Specification FF912. Among other things,
this means that:• The diagnostic category according to NAMUR Recommendation NE107 is transmitted via
the fieldbus in a format that is independent of the manufacturer:– F: Failure– C: Function check– S: Out of specification– M: Maintenance required• The user can change the diagnostic category of the event groups specified in accordancewith the requirements of the particular application.
• Certain events can be separated from their group and handled separately:– e.g. 115: Sensor overpressure– e.g. 715: Sensor over temperature• Additional information and troubleshooting measures are transmitted with the event
message via the fieldbus.

7.9.1

Event groups

The diagnostic events are divided into 16 groups, depending on the source and the severityof the event. A default event category is assigned to each group at the factory. One bit of theassignment parameters belongs to every event group.
Event severity

Default event category

Event source

Bit

Events in this group

Highest severity

Failure (F)Sensor31• 122: F>Sensor not connected• 716: F>Process isolating diaphragm broken

Electronics

30• 110: F>Checksum error in EEPROM: configuration segment• 113: F>ROM failure in transmitter electronic• 121: F>Checksum error in factory segment of EEPROM• 130: F>EEPROM is defect.
• 131: F>Checksum error in EEPROM: min/max segment• 133: F>Checksum error in History EEPROM• 135: F>Checksum error in EEPROM FF segment• 728: F>RAM error• 729: F>RAM error• 736: F>RAM error

Configuration

29• Not used

Process28• Not used

68

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Commissioning

Event severity

Default event category

Event source

Bit

Events in this group

High severity

Check (C)

Sensor27• 101: C>Sensor electronic EEPROM error• 725: C>Sensor connection error, cycle disturbance• 747: C>Sensor software not compatible to electronics

Electronics

26• 703: C>Measurement error• 704: C>Measurement error• 705: C>Measurement error• 737: C>Measurement error• 738: C>Measurement error• 739: C>Measurement error• 742: C>Sensor connection error (upload)• 743: C>Electronic PCB error during initialization• 744: C>Main electronic PCB error• 746: C>Sensor connection error - initializing• 748: C>Memory failure in signal processor

Configuration25• 106: C>Downloading - please wait
• 613: C>Simulation active

Process24• Not used

Event severity

Default event category

Event source

Bit

Events in this group

Low severity

Out of specification(S)

Sensor23• 115: S>Sensor overpressure• 120: S>Sensor low pressure• 715: S>Sensor over temperature• 720: S>Sensor under temperature• 726: S>Sensor temperature error - overrange

Electronics

22• 717: S>Transmitter over temperature• 718: S>Transmitter under temperature

Configuration

21• 701: S>Adjustment outside sensor nominal range• 710: S>Set span too small. Not allowed.
• 727: S>Sensor pressure error - overrange

Process

20• 740: S>Calculation overflow, bad configuration

Endress+Hauser

69Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Event severity

Default
event category

Event source

Bit

Events in this group

Lowest severity

Maintenance required(M)

Sensor19• 745: M>Sensor data unknown

Electronics

18• 134: M>EEPROM lifetime WARNING• 700: M>Last configuration not stored• 702: M>HistoROM data not consistent

Configuration

17• 116: M>Download error, repeat download• 602: M>Linearization curve not monoton• 604: M>Linearization table invalid. Min. 2 points.
• 706: M>Configuration in HistoROM and device not identical.
• 707: M>X-VAL. (TAB_XY_VALUE) of lin. table out of edit limits• 711: M>LRV or URV out of edit limits• 713: M>100% POINT (LEVEL_100_PERCENT_VALUE) level out of edit
limits• 719: M>Y-VALUE (TAB_XY_VALUE) of lin. table out of edit limits• 721: M>ZERO POSITION (LEVEL_OFFSET) level out of edit limits• 722: M>EMPTY CALIB. (SCALE_OUT, EU_0) or FULL CALIB. (SCALE_OUT,
EU_100) out of edit limits• 741: M>TANK HEIGHT (LEVEL_TANK_HEIGHT) out of edit limits• 750: M>Configuration not permitted

Process

16• 730: M>Pmin ALARM WINDOW (PRESSURE_1_USER_LOW_LIMIT)
undershot• 731: M>Pmax ALARM WINDOW (PRESSURE_1_USER_HIGH_LIMIT)
overshot• 732: M>Tmin ALARM WINDOW (TEMPERATURE_1_USER_LOW_LIMIT)
undershot• 733: M>Tmax ALARM WINDOW (TEMPERATURE_1_USER_HIGH_LIMIT)
overshot

7.9.2

Assignment parameters

The event categories are assigned to the event groups via four assignment parameters.
These parameters are located in the RESOURCE (RB2) Block:• FD_FAIL_MAP: for event category Failure (F)• FD_CHECK_MAP: for event category Check (C)• FD_OFFSPEC_MAP: for event category Out of Specification (S)• FD_MAINT_MAP: for event category Maintenance Required (M)Each of these parameters comprises 32 bits with the following meaning:• Bit 0: reserved by the Fieldbus Foundation. Is also set if 1 TRD is not in the AUTO mode.
• Bits 1 to 15: configurable area; certain diagnostic events can be assigned here irrespectiveof the event group they are in. They are not excluded from the event group and their
behavior can be configured individually ( ä 73). In the case of Deltabar S, the following
events can be assigned to the configurable area:– e.g. 115: Sensor overpressure– e.g. 715: Sensor over temperature• Bits 16 to 31: standard area; these bits are permanently assigned to the event groups. If
the bit is set to 1 this event group is assigned to the particular event category.
The factory setting of the assignment parameters is indicated in the following table. In thefactory setting, there is a clear assignment between the event severity and the event
category (e.g. the assignment parameter).

70

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Commissioning

Factory setting of assignment parameters
Standard area
Event severity

Highest severity

High severity

Configurable area

Low severity

Lowest severity

Event source1)

S

E

C

P

S

E

C

P

S

E

C

P

S

E

C

P

Bit

31

30

29

28

27

26

25

24

23

22

21

20

19

18

17

16

15...1

FD_FAIL_MAP

1

1

1

1

0

0

0

0

0

0

0

0

0

0

0

0

0

FD_CHECK_MAP

0

0

0

0

1

1

1

1

0

0

0

0

0

0

0

0

0

FD_OFFSPEC_MAP

0

0

0

0

0

0

0

0

1

1

1

1

0

0

0

0

0

FD_MAINT_MAP

0

0

0

0

0

0

0

0

0

0

0

0

1

1

1

1

0

1)

S: Sensor; E: Electronics; C: Configuration; P: Process

Proceed as follows to change the diagnostic behavior of an event group:
1.

Open the assignment parameter to which the group is currently assigned.

2.

Change the bit of the event group from 1 to 0. If using FieldCare, this is done via theFF912 module by disabling the appropriate check box (see the following example).

3.

Open the assignment parameter to which the group should be assigned.

4.

Change the bit of the event group from 0 to 1. If using FieldCare, this is done via byenabling the appropriate check box (see the following example).

ExampleThe Highest Severity / Electronics group contains the events 131: Checksum error in
EEPROM: min/max segment, among others. These should no longer be categorized as
Failure (F), and should be categorized as Check (C) instead.
1.

Endress+Hauser

In the FieldCare navigation window navigate to Expert Field Device Diagnostic Configuration

71

Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

2.

In the Failure column search for the Highest Severity Electronic group and disable theassociated check box (A). Enable the appropriate check box in the Function column (B).
Please note that the "Accept" button must be pressed to confirm each entry.

(A)

(B)

It must be ensured that the appropriate bit is set in at least one of the assignment
parameters for each event group. Otherwise a category is not transmitted with the event via
the bus. The control system will therefore generally ignore the presence of the event.
The detection of diagnostic events and the transmission of the messages on the bus isconfigured on the FieldCare page Expert Field Device Diagnostic Configuration. The"Mask" column is used for the transmission of the message on the bus. It must be noted thatdevice messages can still be transmitted by polling the active bits in Status 1 and 2. The mask
check box acts as a negative check box, i.e. if a field is selected the associated events are nottransmitted on the bus. The Resource Block must be set to the Auto mode to transmit status
information on the bus.

72

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.9.3

Commissioning

Configurable area

The event category can be individually defined for the following events - irrespective of theevent group to which they are assigned in the factory setting:• 115: Sensor overpressure• 120: Sensor low pressure• 715: Sensor over temperature• 717: Transmitter over temperature• 718: Transmitter under temperature• 720: Sensor under temperature• 726: Sensor temperature error-overrange• 727: Sensor pressure error-overrange• 730: LRV user limits exceeded• 731: URV user limits exceeded• 732: LRV-Temp. user limits exceeded• 733: URV-Temp. user limits exceeded• 740: Calculation Overflow, bad configuration
To change the event category, the event must first be assigned to one of the bits 1 to 15. The
FF912ConfigArea_1 to FF912ConfigArea_15 parameters in the DIAGNOSTIC(TRDDIAG) Block are used for this purpose. The appropriate bit can then be set from 0 to 1
in the desired assignment parameter.
ExampleError 115 "Sensor overpressure" should no longer be categorized as Out of Specification(S), and should be categorized as Function (C) instead.
1.

Endress+Hauser

In the FieldCare navigation window navigate to Expert Field Device Diagnostic Configuration.

73

Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

2.

Select the "Configurable area" tab. In the factory setting, all the bits in the Configurable
Area Bits column have the value not assigned (A).

(A) (B)(C)

3.

Select one of these bits (here Configurable Area Bit 15, for instance) and select the
Sensor overpressure (B) option from the associated picklist. Confirm your choice bypressing "Accept".

4.

Enable the check box for the bit concerned (here: Configurable Area Bit 15) (C).
Confirm your choice by pressing "Accept".

Additional information:
The "Status 1" and "Status 2" tabs indicate whether an event is active.

Changing the error category for Sensor overpressure does not affect an error that alreadyexists. The new category is only assigned if this error occurs again after the change has beenmade. The "Status 1" and "Status 2" tabs indicate whether an event is active.

74

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Commissioning

The "Simulation" tab makes it possible to simulate an event.

Endress+Hauser

75Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.9.4

Transmission of independent event messages on the bus

Event priority
Event messages are only transmitted on the bus if their priority is between 2 and 15. Priority
1 events are displayed but are not transmitted on the bus. Events whose priority is 0 are
ignored. All events are assigned the priority 0 in the factory setting. It is possible to change
the priority individually for the four assignment parameters.
ExampleThe priority of the "Failure" category is to be set to "2" .
1.

In the FieldCare navigation window navigate to Expert Field Device Diagnostics Configuration2.

Select the "Standard area" tab and set the priority to "2" in the "Failure" column (D).

(D)

Suppressing certain events
It is possible to suppress certain events during transmission on the bus using a "mask" checkbox. While these events are still displayed, they are not transmitted as alert objects on thebus. This mask check box can be found in FieldCare under Expert  Field Device Diagnostic
Configuration. The mask check box acts as a negative check box, i.e. if a field is selected
the associated events are not transmitted on the bus.

76

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

7.9.5

Commissioning

Overview of the settings made and the current events

1.

In the FieldCare navigation window navigate to Diagnose Alarm Indication(Polling)

2.

The following overview is displayed:– "Troubleshooting information" if an event has occurred– "Setting made" in the configurable area
– "Current events" in the various categories

7.9.6

Information about the current events

1.

In the FieldCare navigation window navigate to Expert Field Device Diagnostic Actual Alarm Condition

2.

The following overview is displayed:– "Troubleshooting information" if an event has occurred– "FF912 Field Diagnostic Profile" version– "Information about the current events" in the various categories

7.9.7

Setting the status of the flexible alarms

The event category can be individually defined for the following events - irrespective of theevent group to which they are assigned in the factory setting:• 115: Sensor overpressure• 120: Sensor low pressure• 715: Sensor over temperature• 717: Transmitter over temperature• 718: Transmitter under temperature• 720: Sensor under temperature• 726: Sensor temperature error-overrange• 727: Sensor pressure error-overrange• 730: LRV user limits exceeded• 731: URV user limits exceeded• 732: LRV-Temp. user limits exceeded• 733: URV-Temp. user limits exceeded
Endress+Hauser

77Commissioning

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus• 740: Calculation Overflow, bad configuration
To change the assigned status (Bad, Uncertain, Good) of an event, select the desired statusfrom the associated picklist.
ExampleThe status of error 115 "Sensor overpressure" should no longer be indicated as "Uncertain"
and should instead be indicated as "Bad".

78

1.

In the FieldCare navigation window navigate to Expert Field Device Diagnostics 
PV Status Config

2.

All the bits of the Status Select Events have the status "Uncertain" in the factory setting.

3.

Select the "Bad" option for the "Status Select Event 115" row. Press Enter to confirm yourentry.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

8

Maintenance

Maintenance

Keep the pressure compensation and GORE-TEX® filter (1) free from contamination andwater.

1

P01-PMC71xxx-17-xx-xx-xx-001

8.1

Cleaning instructions

Endress+Hauser offer flushing rings as accessories to clean process isolating diaphragms
without taking the transmitters out of the process.
For further information please contact your local Endress+Hauser Sales Center.

8.1.1

PMP75

We recommend you perform CIP (cleaning in place (hot water)) before SIP (sterilization inplace (steam)) for pipe diaphragm seals.
A frequent use of sterilization in place (SIP) will increase the stress on the process isolating
diaphragm. Under unfavorable circumstances in the long term view we cannot exclude thata frequent temperature change could lead to a material fatigue of the process isolating
diaphragm and possibly to a leakage.

8.2

Exterior cleaning

Please note the following points when cleaning the device:• The cleaning agents used should not attack the surface and the seals.
• Mechanical damage to the process isolating diaphragm, e.g. due to pointed objects, must
be avoided.
• Observe degree of protection. See therefor nameplate if necessary ( ä 10).

Endress+Hauser

79Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9

Diagnostics and troubleshooting

9.1

Troubleshooting

9.1.1

General errors

Error

Possible cause

RemedyDevice is not
responding.

The supply voltage does not match thespecifications on the nameplate.

Apply correct voltage.

Supply voltage has incorrect polarity.

Reverse polarity of supply voltage.

No contact between connecting cables andterminals.

Check the contacting of the cables andcorrect if necessary.

Onsite display is set too bright or too dark.

• Set the local display brighter bysimultaneously pressing O and F.
• Set the local display darker bysimultaneously pressing S and F.

No display

Connector for local display is not plugged in Ensure connector is plugged in correctly.
correctly.

Device measuresincorrectly.

9.1.2

Onsite display is defective.

Replace onsite display.

Configuration error.

Check and correct parameter configuration(see below).

Displaying messages:• Onsite display:– The measured value display shows the message with the highest priority. See the"Priority" column.
– The ALARM STATUS parameter shows all the messages present in descending order ofpriority. You can scroll through all the messages pending using the S or O key.
• FieldCare:The DIAGNOSTIC_CODE/DIAGNOSE_CODE parameter displays the message with thehighest priority.
See the "Priority" column.
See also Section 9.6 "Response of outputs to errors".
• Diagnose Transducer Block (FF configuration program):
The DIAGNOSTIC_CODE/DIAGNOSE_CODE parameter displays the message with thehighest priority. See also Section 9.6 "Response of outputs to errors". Every message isalso output as per the FOUNDATION Fieldbus Specification by means of the XD_ERROR
and BLOCK_ERROR parameters in the Pressure, Service and DP Flow Block. Numbers are
given for these parameters in the following table which are explained on 83.
• You can see a list of all the active alarms via the Diagnostic code/ACTUAL_ALARM_INFOS
parameter.
• You can see a list of all the alarms that are no longer active (event log) via the Last Diag.
Code/LAST_ALARM_INFOS parameter.

80

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostics and troubleshooting

9.2

Diagnostic information on local display

9.2.1

Diagnostic message

Faults detected by the self-monitoring system of the measuring device are displayed as adiagnostic message in alternation with the measured value display.
Error categories

F
C
S
M

"Failure"
A device error has occurred. The measured value is no longer valid.
"Check"The device is in the service mode (e.g. during a simulation) or is performing a self-check.
"Out of specification"
The device is being operated:• Outside its technical specifications (e.g. during warmup or cleaning processes).
• Outside the parameter configuration undertaken by the user (e.g. pressure outside of
nominal operating range)"Maintenance required"
Maintenance is required. The measured value is still valid.

Diagnostic event and event text
The fault can be identified using the diagnostic event. The event text helps you by providing
information about the fault.
Diagnostic event
Status signal

ExampleEvent number

Event text with category
information

E115

S>Sensor overpressure

3-digit number• If the device detects a defect in the onsite display during initialization, special error
messages are generated. For the error messages, see  ä 81, Section 9.2.2 "Onsite
display error messages".
• For support and further information, please contact Endress+Hauser Service.
• See also Section 9.8, Section 9.9 and Section 9.10.
• If the category of a diagnostic event is being changed, an empty field might be displayed
instead of "F, C, S, M".

9.2.2

Onsite display error messages

If the device detects a defect in the onsite display during initialization, the following error
messages can be displayed:

Endress+Hauser

81Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Message

MeasureInitialization, VU Electr. Defect A110
Initialization, VU Electr. Defect A114
Initialization, VU Electr. Defect A281
Initialization, VU Checksum Err. A110
Initialization, VU Checksum Err. A112
Initialization, VU Checksum Err. A171

Exchange onsite display.

9.3

Diagnostic event in the operating tool

If a diagnostic event is present in the operating tool, the status signal appears in the top left
statusarea along with the corresponding symbol for event behavior in accordance with NAMUR NE
107:• Failure (F)• Check (C)• Maintenance required (M)• Out of specification (S)P01-xMx7xxxx-05-xx-xx-en-008

Calling up remedy information

82

1.

Navigate to the "Diagnostics" menu. The diagnostic event along with text about theevent is displayed in the "Actual diagnostics" parameter.

2.

On the right in the display area, hover the cursor over the "Actual diagnostics"
parameter. A tool tip with remedy information for the diagnostic event appears.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.4

Diagnostics and troubleshooting

Diagnostic messages in the DIAGNOSTIC Transducer
Block (TRDDIAG)• The Actual Diagnostics parameter shows the message with the highest priority. Every
message is also displayed as per the FOUNDATION Fieldbus Specification by means of the
XD_ERROR and BLOCK_ERROR parameters.
• You can view the active alarm with the highest priority via the Diagnosis parameter.
• You can view the last alarm that is no longer active via the Last Diagnosis parameter.

Diagnostic CodeXD_ERROR Value Bit

XD_ERROR Text

BLOCK_ERROR Value Bit

BLOCK_ERROR Text

PRIMARY_VALUE(Status is set according toOperating Mode)

PRIMARY_VALUE_TYPE(Operating Mode)

TRANSDUCER Status
Propagation(Impacted CHANNEL
selection)

Explanation of XD_ERROR, BLOCK_ERROR and response of
outputs

Error type

9.4.1

Alarm

747

17

General Error

0

Other

BAD_SENSOR_FAILURE

Pressure, Level, Flow

All

707

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Level

Primary Value(1)

711

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Level, Flow

Primary Value(1)Totalizer 1 (6)

713

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Level

Primary Value(1)

721

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Level

Primary Value(1)

722

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Level

Primary Value(1)

723

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Flow

Primary Value(1)Totalizer 1 (6)

741

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Level

Primary Value(1)

719

19

Configuration
Error

0

Other

BAD_NON_SPECIFIC

Level

Primary Value(1)

750

18

Calibration Error

0

Other

BAD_NON_SPECIFIC

Pressure, Level, Flow

Primary Value(1)Pressure(3)Maximum Pressure(4)Counter P > Pmax(5)Totalizer 1 (6)

122

20

Electronics Failure

7

Sensor Failure BAD_SENSOR_FAILURE

Pressure, Level, Flow

All

101

20

Electronics Failure

0

Other

BAD_SENSOR_FAILURE

Pressure, Level, Flow

All

716

20

Electronics Failure

0

Other

BAD_SENSOR_FAILURE

Pressure, Level, Flow

All

725

20

Electronics Failure

0

Other

BAD_SENSOR_FAILURE

Pressure, Level, Flow

All

704

20

Electronics Failure

7

Sensor Failure BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

703

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

705

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

737

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

738

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

739

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

742

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

744

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

743

20

Electronics Failure

7

Sensor Failure BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

748

20

Electronics Failure

7

Sensor Failure BAD_DEVICE_FAILURE

Pressure, Level, FlowAll

Endress+Hauser

83

Diagnostic CodeXD_ERROR Value Bit

XD_ERROR Text

BLOCK_ERROR Value Bit

BLOCK_ERROR Text

PRIMARY_VALUE(Status is set according toOperating Mode)

PRIMARY_VALUE_TYPE(Operating Mode)

TRANSDUCER Status
Propagation(Impacted CHANNEL
selection)

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Error type

Diagnostics and troubleshooting

Alarm

113

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

728

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

729

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

736

20

Electronics Failure

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

110

23

Data Integrity
Error

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

130

23

Data Integrity
Error

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

131

23

Data Integrity
Error

0

Other

GOOD

Pressure, Level, Flow

None

132

23

Data Integrity
Error

0

Other

BAD_DEVICE_FAILURE

Flow

Totalizer 1 (6)

133

23

Data Integrity
Error

0

Other

GOOD

Pressure, Level, Flow

None

135

23

Data Integrity
Error

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

121

23

Data Integrity
Error

0

Other

BAD_DEVICE_FAILURE

Pressure, Level, Flow

All

115

17

General Error

0

Other

– BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

Primary Value(1)Pressure(3)Maximum Pressure(4)Counter P > Pmax(5)Totalizer 1 (6)

120

17

General Error

0

Other

– BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

Primary Value(1)Pressure(3)Maximum Pressure(4)Counter P > Pmax(5)Totalizer 1 (6)

717

17

General Error

0

Other

– BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

All

718

17

General Error

0

Other

– BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

All

720

17

General Error

0

Other

– BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

Sensor Temperature(2)

715

17

General Error

7

Sensor Failure – BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

Sensor Temperature(2)

726

20

Electronics Failure

7

Sensor Failure – BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

All

740

20

Electronics Failure

7

Sensor Failure – BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

All

727

20

Electronics Failure

7

Sensor Failure – BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, FlowAll

Alarm/
Warning

84

Endress+Hauser

BLOCK_ERROR Value Bit

BLOCK_ERROR Text

PRIMARY_VALUE(Status is set according toOperating Mode)

PRIMARY_VALUE_TYPE(Operating Mode)

TRANSDUCER Status
Propagation(Impacted CHANNEL
selection)

730

19

Configuration
Error

0

Other

GOOD

Pressure, Level, Flow

None

731

19

Configuration
Error

0

Other

GOOD

Pressure, Level, Flow

None

732

19

Configuration
Error

0

Other

GOOD

Pressure, Level, Flow

None

733

19

Configuration
Error

0

Other

GOOD

Pressure, Level, Flow

None

106

17

General Error

0

Other

STATUS_UNCERTAIN

Pressure, Level, Flow

All

134

17

General Error

0

Other

GOOD

Pressure, Level, Flow

None

116

17

General Error

0

Other

– BAD_NON_SPECIFIC– STATUS_UNCERTAIN– GOOD

Pressure, Level, Flow

All

701

17

General Error

0

Other

UNCERTAIN_CONFIG_ERROR Pressure, Level, Flow

All

745

17

General Error

0

Other

STATUS_UNCERTAIN

Pressure, Level, Flow

All

613

17

General Error

0

Other

UNCERTAIN_SIM

Pressure, Level, Flow

Primary Value(1)Maximum Pressure(4)Counter P > Pmax(5)Totalizer 1 (6)

702

17

General Error

0

Other

GOOD

Pressure, Level, Flow

None

710

18

Calibration Error

0

Other

GOOD

Pressure, Level, Flow

None

602

19

Configuration
Error

0

Other

UNCERTAIN_CONFIG_ERROR Level

Primary Value(1)

604

19

Configuration
Error

0

Other

UNCERTAIN_CONFIG_ERROR Level

Primary Value(1)

746

20

Electronics Failure

0

Other

STATUS_UNCERTAIN

Pressure, Level, Flow

All

102

23

Data Integrity
Error

0

Other

GOOD

Pressure, Level, Flow

Maximum Pressure(4)Counter P > Pmax(5)

700

23

Data Integrity
Error

0

Other

STATUS_UNCERTAIN

Pressure, Level, Flow

All

706

23

Data Integrity
Error

0

Other

GOOD

Pressure, Level, Flow

None

Endress+Hauser

XD_ERROR Text

XD_ERROR Value Bit

WarningDiagnostics and troubleshooting

Diagnostic CodeError type

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

85

Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.5

Overview of diagnostic events

9.5.1

Failure (F)Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

MeasurePriority

101

F>Sensor electronic
EEPROM error

20– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.) Thismessage normally only
appears briefly.

– Wait a few minutes.
– Restart the device. Perform
reset (Code 62).
– Block off electromagnetic
effects or eliminate source ofdisturbance.

19– Sensor defect.

– Replace sensor.

– The supply voltage is
disconnected when writing.

– Reestablish supply voltage.
Perform reset if necessary(code 7864) and recalibrate
the device.

– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.)

– Block off electromagnetic
effects or eliminate sources ofdisturbance.

– Main electronics defect.

– Replace main electronics.

110

Alarm

Alarm

F>Checksum error in
EEPROM:configuration segment

23

0

0

6

113

Alarm

F>ROM failure intransmitter electronic

200– Main electronics defect.

– Replace main electronics.

1

121

Alarm

F>Checksum error infactory segment of
EEPROM230– Main electronics defect.

– Replace main electronics.

5

122

Alarm

F>Sensor notconnected

20

7– Cable connection sensor –main electronics
disconnected.

– Check cable connection andrepair if necessary.

14– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.)

– Block off electromagnetic
effects or eliminate source ofdisturbance.

– Main electronics defect.

– Replace main electronics.

– Sensor defect.

– Replace sensor.

130

Alarm

F>EEPROM is defect.

230– Main electronics defect.

– Replace main electronics.

11

131

Alarm

F>Checksum error in
EEPROM: min/max
segment230– Main electronics defect.

– Replace main electronics.

9

133

Alarm

F>Checksum error in
History EEPROM

23

0– An error occurred whenwriting.

– Perform reset (code 7864)
and recalibrate the device.

8– Main electronics defect.

– Replace main electronics.

135

Alarm

F>Checksum error in
EEPROM FF segment

230– Main electronics defect.

– Replace main electronics.

703

Alarm

F>Measurement error

200– Fault in the main electronics.

– Briefly disconnect device from 24
the power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from 23
the power supply.

– Main electronics defect.

– Replace main electronics.

– Sensor defect.

– Replace sensor.
– Reduce pressure.

705

716

86

Alarm

Alarm

F>Measurement error

F>Process isolating
diaphragm broken

20

20

0

0

10

26

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostics and troubleshooting

Diagnostic Error
Coderesponse

Message/description

MeasurePriority

725

F>Sensor connection
20
error, cycle disturbance– Electromagnetic effects are
greater than specifications in
the technical data. (See
Section 10.)– Setscrew loose.

– Block off electromagnetic
effects or eliminate source ofdisturbance.

27– Sensor or main electronics
defect.

– Replace sensor or mainelectronics.

– Fault in the main electronics.

– Briefly disconnect device from 2
the power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from 3
the power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from 4
the power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from 22
the power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from 21
the power supply.

– Main electronics defect.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from 25
the power supply.

– Main electronics defect.

– Replace main electronics.

– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.) Thismessage normally only
appears briefly.

– Wait a few minutes.
– Perform reset (code 7864)
and recalibrate the device.

– Cable connection sensor –main electronics
disconnected.

– Check cable connection andrepair if necessary.

– Sensor defect.

– Replace sensor.

– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.) Thismessage normally only
appears briefly.

– Wait a few minutes.
– Restart the device. Perform
reset (Code 62).

– Main electronics defect.

– Replace main electronics.

– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.)– Restart the device. Perform
reset (Code 62).
– Block off electromagnetic
effects or eliminate source ofdisturbance.

– Main electronics defect.

– Replace main electronics.

– Sensor does not suit thedevice (electronic sensor
nameplate).

– Replace sensor with a suitable 18
sensor.

728

729

736

737

738

739

742

743

744

747

Alarm

Alarm

Alarm

Alarm

Alarm

Alarm

Alarm

Alarm

Alarm

Alarm

Alarm

Endress+Hauser

F>RAM error

F>RAM error

F>RAM error

F>Measurement error

F>Measurement error

F>Measurement error

F>Sensor connection
error (upload)

F>Electronic PCB error
during initialization

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

20

20

20

20

20

20

20

20

F>Main electronic PCB 20
error

F>Sensor software notcompatible toelectronics

17

0

0

0

0

0

0

0

0

7

0

0

– Retighten setscrew with 1 Nm
(0.74 lbf ft) (see Chap. 4.4.8).

20

15 /
16

12

87

Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

MeasurePriority

748

F>Memory failure insignal processor

20– Electromagnetic effects are
greater than specifications in
the technical data.
( See Section 10.)

– Block off electromagnetic
effects or eliminate source ofdisturbance.

17– Main electronics defect.

– Replace main electronics.

Alarm

9.5.2

7

Check (C)Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

MeasurePriority

106

Warning

C>Downloading please wait

170

– Downloading.

– Wait for download tocomplete.

61

602

Warning

C>Linearization curve
not monoton

190– The linearization table is not
monotonic increasing.

– Add to or correct linearization 67
table. Then accept
linearization table again.

604

Warning

C>Linearization table
invalid. Min. 2 points.

19

0

Note! There is no min. span for the y-points as of software version"03.00.00".
– The linearization table has
fewer than 2 points.

68
1. Add to linearization table.
Perform the linearization again
if necessary.
2. Correct the linearization table
and accept it again.

613

Warning

C>Simulation active

170– Simulation is switched on, i.e. – Switch off simulation.
the device is not measuring atpresent.

70

701

Warning

C>Adjustment outside
sensor nominal range

170– The adjustment carried out
would cause a breach of thenominal sensor range.

– Carry out calibration again.

704

Alarm

C>Measurement error

207– Fault in the main electronics.

– Briefly disconnect device from 13
the power supply.

– Main electronics defect.

– Replace main electronics.

63

707

Alarm

C>X-VAL.
(TAB_XY_VALUE) of
lin. table out of edit
limits.

18

0– At least one X-VALUE(TAB_XY_VALUE) in thelinearization table is either
below the value for
SCALE_IN, EU_0/HYDR.
PRESS MIN. or LINEAR_LEVEL_MIN/MIN. LEVEL orabove the value for
SCALE_IN, EU_100/HYDR.
PRESS. MAX. or
LINEAR_LEVEL_MAX/LEVEL MAX.

45
– Perform calibration again(See Operating Instructions
BA00303P, parameter
description, Section 5).

710

Warning

B>Set span too small.
Not allowed.

18

0– Values for calibration (e.g.
lower range value and upper
range value) are too close
together.

– Adjust calibration to suitsensor (see Operating
Instructions BA00303P,
parameter description,
CAL_MIN_SPAN/MINIMUM
SPAN parameter).

– The sensor was replaced andthe customer-specificconfiguration does not suit
the sensor.

– Adjust calibration to suitsensor.
– Replace sensor with a suitablesensor.

– Unsuitable download carried
out.

– Check configuration andperform download again.

88

60

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostics and troubleshooting

Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

Measure711

C>LRV or URV out ofedit limits

18

Alarm

0

– Reset the LRV and/or URV to– Lower range value and/or
suit the sensor. Pay attention
upper range value undershootto position adjustment.
or overshoot the sensor range
limits.
– The sensor was replaced andthe customer-specificconfiguration does not suit
the sensor.

– Reset the LRV and/or URV tosuit the sensor. Pay attention
to position adjustment.
– Replace sensor with a suitablesensor.

– Unsuitable download carried
out.

– Check configuration andperform download again.

Priority
37

713

Alarm

C>100% POINT(LEVEL_100_PERCEN
T_VALUE) level out ofedit limits

180– The sensor was replaced.

– Carry out calibration again.

719

Alarm

C>Y-VAL(TAB_XY_VALUE) of
lin. table out of edit
limits19

0– At least one Y-VALUE(TAB_XY_VALUE) in thelinearization table is below
the SCALE_OUT, EU_0/
TANK CONTENT MIN. orabove the SCALE_OUT,
EU_100/TANK CONTENT
MAX.

47– Carry out calibration again.
(See Operating Instructions
BA00303P, parameter
description, Section 5).

721

Alarm

C>ZERO POSITION(LEVEL OFFSET) levelout of edit limits

180– LEVEL MIN(LINEAR_LEVEL_ MIN) or
LEVEL MAX
(LINEAR_LEVEL_MAX) has
been changed.

– Perform reset (code 2710)
and recalibrate the device.

722

Alarm

C>EMPTY CALIB.
(SCALE_OUT, EU_0)
or FULL CALIB.
(SCALE_OUT,
EU_100) out of edit
limits18

0– LINEAR_ LEVEL_MIN/LEVEL – Perform reset (code 2710)
and recalibrate the device.
MIN or
LINEAR_LEVEL_MAX/LEVEL MAX has beenchanged.

49/50

723

Alarm

C>MAX. FLOW(SCALE_OUT,
EU_100) out of edit
limits18

0– FLOW_TYPE/FLOW-MEAS.
TYPE has been changed.

– Carry out calibration again.

51

741

Alarm

18
C>TANK HEIGHT(LEVEL_TANK_HEIGHT) out of edit limits

0– LINEAR_LEVEL_MIN/LEVELMIN or
LINEAR_LEVEL_MAX/LEVEL MAX has beenchanged.

– Perform reset (code 2710)
and recalibrate the device.

52

746

Warning

C>Sensor connection
error - initializing0– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.) Thismessage normally only
appears briefly.

– Wait a few minutes.
– Restart the device. Perform
reset (Code 7864).
– Block off electromagnetic
effects or eliminate source ofdisturbance.

28– Overpressure or low pressure
present.

– Reduce or increase pressure.

Endress+Hauser

20

46

48

89

Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

MeasurePriority

750

C>Configuration not
permitted

18– Check configuration.
– Perform reset (code 7864)
and recalibrate the device.

53

90

Warning0– By means of the operation
profile, options were selected
for the configuration of thedevice but the options do notsuit one another. For
example, if the option "1"(linearization table) was
selected for LIN_TYPE andthe unit "1347 (m3/s)" wasselected for
PRIMARY_VALUE_UNIT.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.5.3

Diagnostics and troubleshooting

Maintenance required (M)Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

MeasurePriority

102

Warning

M>Checksum error in
EEPROM: peakhold
segment

23

0– Replace main electronics.
– Main electronics defect.
Correct measurement cancontinue as long as you do notneed the peak hold indicatorfunction.

62

116

Warning

M>Download error,
repeat download

170– The file is defect.

– Use another file.

38– During the download, thedata are not correctlytransmitted to the processor,
e.g. because of open cableconnections, spikes (ripple)
on the supply voltage orelectromagnetic effects.

– Check cable connection PC –transmitter.
– Block off electromagnetic
effects or eliminate sources ofdisturbance.
– Perform reset (code 7864)
and recalibrate the device.
– Repeat download.

134

Warning

M>EEPROM lifetime
WARNING170– Writing too often to
EEPROM.

– Reduce write accessing to
EEPROM.

65

700

Warning

M>Last configuration
not stored

23

0– An error occurred whenwriting or reading
configuration data or the
power supply was
disconnected.

– Perform reset (code 7864)
and recalibrate the device.

63– Main electronics defect.

– Replace main electronics.

– Data were not writtencorrectly to the HistoROM,
e.g. if the HistoROM wasdetached during the writing
process.

– Repeat upload.
– Perform reset (code 7864)
and recalibrate the device.

– HistoROM does not have anydata.

– Copy suitable data to theHistoROM. ( ä 50,
Section 6.6.1 "Copying
configuration data".)

702

706

Warning

WarningEndress+Hauser

M>HistoROM data notconsistent

M>Configuration in
HistoROM and device
not identical

17

23

0

0

64

– Configuration (parameters) in – Copy data from the device to 69
the HistoROM. ( ä 50,
the HistoROM and in the
Section 6.6.1 "Copying
device not identical.
configuration data".)– Copy data from the HistoROM
to the device. ( ä 50,
Section 6.6.1 "Copying
configuration data".) The
message remains if theHistoROM and the device
have different software
versions. The message
disappears if you copy thedata from the device to theHistoROM.
– Device reset codes such as 1
or 40864 do not have any
effect on the HistoROM. Thatmeans that if you perform areset, the configurations in
the HistoROM and in thedevice can be different.

91Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

Measure740

M>Calculation
overflow, badconfiguration

20

29– Check configuration andrecalibrate the device ifnecessary.
– Select a device with a suitablemeasuring range.

Alarm/
warning

7– Level measuring mode: themeasured pressure has
undershot the value for
SCALE_IN, EU_0/HYDR.
PRESS. MIN. or overshot thevalue for SCALE_IN,
EU_100/HYDR. PRESS MAX.

Priority– Check configuration and– Level measuring mode: therecalibrate the device if
measured level has undershot
necessary (see Operating
the value for LEVEL MIN or
Instructions BA00303P,
overshot the value for LEVEL
parameter description, LEVELMAX
MIN. parameter).

745

Warning

M>Sensor data
unknown

9.5.4

17

0

– Sensor does not suit thedevice (electronic sensor
nameplate). Device continues
measuring.

– Replace sensor with a suitable 66
sensor.

Out of specification (S)Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

MeasurePriority

115

S>Sensor overpressure

17– Overpressure present.

– Reduce pressure until
message disappears.

31– Sensor defect.

– Replace sensor.

– Pressure too low.

– Increase pressure until
message disappears.

– Sensor defect.

– Replace sensor.

120

715

Alarm/
warning

Alarm/
warning

Alarm/
warning

S>Sensor low pressure

S>Sensor overtemperature

17

17

0

0

7

– The temperature measured in – Reduce process temperature/
ambient temperature.
the sensor is greater than theupper nominal temperatureof the sensor. (See also
Operating Instructions
BA00303P, parameter
description for
TEMPERATURE_1_SENSOR_LIMIT_HIGH/
Tmax SENSOR or these
Operating Instructions)– Unsuitable download carried
out.

717

Alarm/
warning

S>Transmitterover temperature

17

0

34

– Check configuration andperform download again.

– The temperature measured in – Reduce ambient temperature. 36
the electronics is greater thanthe upper nominal
temperature of the electronics(+88 °C (+190 °F)).
– Unsuitable download carried
out.

92

32

– Check configuration andperform download again.

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostics and troubleshooting

Diagnostic Error
Coderesponse

Message/description

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

Measure718

S>Transmitter
under temperature

17

Alarm/
warning0– The temperature measured in – Increase ambient
temperature. Insulate device
the electronics is smaller thanif necessary.
the lower nominal
temperature of the electronics(–43 °C(-45 °F)).
– Unsuitable download carried
out.

720

726

727

730

Alarm/
warning

Alarm/
warning

Alarm/
warning

Alarm/
WarningEndress+Hauser

S>Sensorunder temperature

S>Sensor temperatureerror - overrange

S>Sensor pressure
error - overrange

S>Pmin ALARM
WINDOW(PRESSURE_1_USER_
LOW_LIMIT)
undershot

17

20

20

19

0

7

7

0

Priority37– Check configuration andperform download again.

– The temperature measured in – Increase processtemperature/ambient
the sensor is lower than thetemperature.
lower nominal temperature of
the sensor (see Operating
Instructions BA00303P,
parameter description),
TEMPERATURE_1_SENSOR_LIMIT_LOW/
Tmin SENSOR parameter).
– Unsuitable download carried
out.

– Check configuration andperform download again.

– Loose connection at sensor
cable.

– Wait a short period of time
and tighten the connection, or
avoid loose connection.

– Electromagnetic effects are
greater than specifications in
the technical data.
( See Section 10.)

– Block off electromagnetic
effects or eliminate source ofdisturbance.

– Process temperature isoutside permitted range.

– Check temperature present,
reduce or increase if
necessary.

– Sensor defect.

– If the process temperature is
within the permitted range,
replace sensor.

– Electromagnetic effects are
greater than specifications in
the technical data.
(See Section 10.)

– Block off electromagnetic
effects or eliminate source ofdisturbance.

– Pressure is outside permitted
range.

– Check pressure present,
reduce or increase if
necessary.

– Sensor defect.

– If the pressure is within thepermitted range, replacesensor.

– Pressure measured value has
undershot the value specified
for the PRESSURE_1_USER_
LOW_LIMIT/Pmin ALARM
WINDOW parameter.

– Check system/pressure
measured value.
– Change value for
PRESSURE_1_
USER_LOW_LIMIT/
Pmin ALARM WINDOW if
necessary.
(See also Operating
Instructions BA00303P,
parameter description orthese Operating Instructions)– Loose connection at sensor
cable.

– Wait a short period of time
and tighten the connection, or
avoid loose connection.

35

33

30

55

93

Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Diagnostic Error
Coderesponse

Message/description

731

Alarm/
Warning

19
S>Pmax ALARM
WINDOW(PRESSURE_1_UER_
HIGH_LIMIT) overshot

0– Pressure measured value has
overshot the value specified
for the PRESSURE_1_USER_HIGH_LIMIT/Pmax ALARM
WINDOW parameter.

732

Alarm/
Warning

S>Tmin ALARM
WINDOW(TEMPERATURE_1_
USER_LOW_LIMIT)
undershot

19

0

733

94

Alarm/
Warning

S>Tmax ALARM
WINDOW(TEMPERATURE_1_
USER_HIGH_LIMIT)
overshot

XD_
BLOCK_ Cause
ERROR
ERROR
Value Bit Value Bit

190

MeasurePriority– Check system/pressure
measured value.
– Change value for
PRESSURE_1_USER_ HIGH_
LIMIT/Pmax ALARM
WINDOW if necessary.
(See also Operating
Instructions BA00303P,
parameter description orthese Operating Instructions)54– Temperature measured value – Check system/temperature
measured value.
has undershot the value– Change value for
specified for the
TEMPERATURE_1_
TEMPERATURE_1_USER_LO
USER_LOW_LIMIT/
W_LIMIT/Tmin ALARM
Tmin ALARM WINDOW if
WINDOW parameter.
necessary.
(See also Operating
Instructions BA00303P,
parameter description orthese Operating Instructions)57– Loose connection at sensor
cable.

– Wait a short period of time
and tighten the connection, or
avoid loose connection.

– Temperature measured value
has overshot the value
specified for the
TEMPERATURE_1_USER_HI
GH_LIMIT/Tmax ALARM
WINDOW parameter.

– Check system/temperature
measured value.
– Change value for
TEMPERATURE_1_
USER_HIGH_LIMIT/
Tmax ALARM WINDOW if
necessary.
(See also Operating
Instructions BA00303P,
parameter description orthese Operating Instructions)56

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.6

Diagnostics and troubleshooting

Response of outputs to errors

The device makes a distinction between the output behavior "Alarm", "Warning" and "Error".
 See the following table and  ä 81, Section 9.2 "Diagnostic information on local display".
Certain problems can be assigned the "GOOD" status via FF communication, see Section 9.4.1.
OutputA (Alarm)

W (Warning)

FOUNDATION
Fieldbus

The process variable in question istransmitted with the status BAD.

Device continues measuring. The process For this error, you can enter whether thevariable in question is transmitted with device should react as in the event of analarm or as in the event of a warning.
the status UNCERTAIN.
See appropriate column for "Alarm" or"Warning"(see Operating Instructions
BA00303P, parameter description),
REACTION_ON_ ALARM_NR/SELECTALARM TYPE parameter).
The GOOD status can also be assigned tothe individual error via the parameters
FF912_STATUS_SELECT_1 toFF912_STATUS_SELECT _131.

Onsite display– The measured value and message are – The measured value and message aredisplayed alternately
displayed alternately-symbol– Measured value display:-symbol is – Measured value display:
flashes.
permanently displayed.

– The measured value and message aredisplayed alternately– Measured value display: seecorresponding "Alarm" or "Warning"columnMessage display:
Message display– A + 3-digit number such as A122 and – W + 3-digit number such as W613
and– Description– Description

Message display:– E + 3-digit number such as E731 and– Description

In the event of an alarm, the ALARM
STATUS/ALARM_STATUS 1) parameter
displays a 3-digit number such as 122
for "Sensor connection error, incorrectdata".

In the case of an error, the ALARM
STATUS/ALARM_STATUS parameter
displays a 3-digit number such as 731
for "Pmax ALARM WINDOW undershot".

Remote operation(FF configuration
program/FieldCare)1)

In the case of a warning, the ALARM
STATUS/ALARM_STATUS parameter
displays a 3-digit number such as 613
for "Simulation is active".

E (Error: Alarm/Warning)

FF configuration program: Diagnostic Transducer Block. Menu path FieldCare: OPERATING MENU  MESSAGESEndress+Hauser

95

Diagnostics and troubleshooting

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.6.1

Analog Input Block

If the Analog Input Block receives an input or simulation value with the status BAD, theAnalog Input Block uses the failsafe mode defined in the FSAFE_TYPE1 parameter.
The following options are available by means of the FSAFE_TYPE parameter:• Last Good Value
The last valid value is used for further processing with the status UNCERTAIN.
• Fail SafeValue
The value specified by means of the FSAFE_VALUE1 parameter is used for further
processing with the status UNCERTAIN.
• Wrong Value
The current value is used for further processing with the status BAD.
Factory setting:• FSAFE_TYPE: FsafeValue• FSAFE_VALUE: 0
The failsafe mode is also activated if the "Out of service" option was selected by means of the
MODE_BLK parameter, "Target" element.
1 These parameters are not available by means of the FieldCare operating program.

9.7

Confirming messages

Depending on the settings for the ALARM_HOLD_ON_TIME/ALARM DISPL. TIME andACKNOWLEDGE_ALARM_MODE/ACK. ALARM MODE parameters, the following
measures should be taken to clear a message:
Settings 1)

Measures– Rectify the cause of the message (see also Section 9.5).
– ALARM_HOLD_ON_TIME//
ALARM DISPL. TIME = 0 s– ACKNOWLEDGE_ALARM_MODE/ACK. ALARM MODE = off– Rectify the cause of the message (see also Section 9.5).
– ALARM_HOLD_ON_TIME/– Wait for the alarm display time to elapse.
ALARM DISPL. TIME > n s– ACKNOWLEDGE_ALARM_MODE/ACK. ALARM MODE = off– Rectify the cause of the message (see also Section 9.5).
– ALARM_HOLD_ON_TIME/– Confirm message using ACKNOWLEDGE_ALARM/ACK. ALARM
ALARM DISPL. TIME = 0 s
parameter.
– ACKNOWLEDGE_ALARM_MODE/ACK. ALARM MODE = on
– Rectify the cause of the message (see also Section 9.5).
– ALARM_HOLD_ON_TIME/– Confirm message using ACKNOWLEDGE_ALARM/ACK. ALARM
ALARM DISPL. TIME > n s
parameter.
– ACKNOWLEDGE_ALARM_MODE– Wait for the alarm display time to elapse. If a message appears and the/ACK. ALARM MODE = onalarm display time elapses before the message has been acknowledged,
the message is cleared once it has been acknowledged.
1)96

FF configuration program: The parameters are in the Diagnostic Transducer Blocks.
FieldCare: Menu path for ALARM DISPL. TIME and ACK. ALARM MODE: OPERATING MENU DIAGNOSTICS MESSAGESEndress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.8

Diagnostics and troubleshooting

Repair

The Endress+Hauser repairs concept provides for measuring devices to have a modular
design and also that the customer may carry out repairs (see  ä 97 "Spare Parts" on ).
• For certified devices, please consult the "Repair of Ex-certified devices" section.
• For more information on service and spare parts contact the Endress+Hauser Service.
( See www.endress.com/worldwide.)

9.9!

Repair of Ex-certified devices

WARNING

Limitation of electrical safety due to incorrect connection!
Risk of explosion!
When repairing Ex-certified devices, please note the following:• Only specialist personnel or Endress+Hauser may undertake repairs of certified devices.
• Relevant standards, national hazardous area regulations and Safety Instructions andCertificates must be observed.
• Only genuine Endress+Hauser spare parts may be used.
• When ordering spare parts, please check the device designation on the nameplate.
Identical parts may only be used as replacements.
• Electronic inserts or sensors already in use in a standard instrument may not be used asspare parts for a certified device.
• Carry out repairs according to the instructions. After repairs, the device must fulfill therequirements of the specified individual tests.
• A certified device may only be converted into another certified variant by Endress+Hauser.
• All repairs and modifications must be documented.

9.10

Spare Parts• Some replaceable measuring device components are identified by means of a spare part
nameplate. This contains information about the spare part.
• All the spare parts for the measuring device along with the order code are listed In the
W@M Device Viewer (www.endress.com/deviceviewer) and can be ordered. If available,
users can also download the associated Installation Instructions.
Measuring device serial number:• Located on the device and spare part nameplate.
• Can be read out via the "DEVICE SERIAL No" parameter in the "TRANSMITTER DATA"
submenu.

9.11

ReturnThe measuring device must be returned if repairs or a factory calibration are required, or if
the wrong measuring device has been ordered or delivered. According to legal regulations,
Endress+Hauser, as a ISO-certified company, is required to follow certain procedures whenhandling returned products that are in contact with process fluids.
To ensure swift, safe and professional device returns, please read the return procedures andconditions on the Endress+Hauser website at www.services.endress.com/return-material.

9.12

Disposal

When disposing, separate and recycle the device components based on the materials.

Endress+Hauser

97

Technical data

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

9.13
Date

03.2005

Software version

02.00.zz

Software history

Changes to the software

Original software.
Compatible with:– ToF Tool Field Tool Package, version 2.04 or higher

08.2008

01.201303.00.zz

04.00.zz

Compatible with:– FieldCare version 2.15.00

FF912 Field Diagnostic Profile Integration

10

Documentation
Operating Instructions

Description of
Device Functions

BA302P/00/EN/04.05
52027501

BA303P/00/EN/04.05
52027502

BA302P/00/EN/11.05
71009599

BA303P/00/EN/04.05
52027502

BA302P/00/EN/10.07
71043307

BA303P/00/EN/04.05
52027502

BA302P/00/EN/12.07
71043307

BA303P/00/EN/04.05
52027502

BA302P/00/EN/05.08
71071782

BA303P/00/EN/05.08
71062794

BA302P/00/EN/08.08
71077547

BA303P/00/EN/08.08
71076564

BA302P/00/EN/06.09
71095438

BA303P/00/EN/06.09
71095456

BA302P/00/EN/05.10
71111799

BA303P/00/EN/05.10
71111814

BA00302P/00/EN/13.11
71139784

BA00303P/00/EN/13.11
71139808

BA00302P/00/EN/14.12
71161903

BA00303P/00/EN/13.11
71139808

BA00302P/00/EN/15.13 BA00303P/00/EN/14.13
71207386
71207399
BA00302P/00/EN/16.14
71254313

BA00303P/00/EN/15.14
71254484

BA00302P/00/EN/17.14
71270366

BA00303P/00/EN/16.14
71270408

BA00302P/00/EN/18.16
71335667

BA00303P/00/EN/17.16
71336364

Technical data

For technical data, please refer to Technical Information Cerabar S TI00383P.

98

Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Index
A

Assignment of Transducer Blocks (CHANNEL) . . . . . . . 39

Menu structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 46
Methods . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45

B

N

Block configuration, delivery status . . . . . . . . . . . . . . . . 38
Block model, Cerabar S . . . . . . . . . . . . . . . . . . . . . . . . . . . 36

C

Cable specification . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27
CHANNEL-Parameter . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Current consumption . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27

D

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Device addressing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
Device identification . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
Diaphragm seals, installation instructions . . . . . . . . . . . 17
Diaphragm seals, vacuum application. . . . . . . . . . . . . . . 18
Display. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29

EElectrical connection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
Error messages. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81

F

Factory setting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
FOUNDATION Fieldbus system architecture . . . . . . . . . 34

G

Grounding. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27

H

Hazardous area . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Heat insulation. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20
HistoROM/M-DAT . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49

I

Incoming acceptance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Index tables . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 41
Installation instructions for devices with diaphragm seals
17
Installation instructions for devices without diaphragmseals. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

K

Keys, onsite, function . . . . . . . . . . . . . . . . . . . . . . . . . 32–33
Keys, position . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 31

L

Level measurement . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 63
Level measurement, Quick Setup menu . . . . . . . . . . . . . 65
Level Quick Setup menu . . . . . . . . . . . . . . . . . . . . . . . . . . 65
Locking operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 52

M

Measuring arrangement for level measurement. . . . . . 17
Measuring arrangement for pressure measurement . 15–
16
Endress+Hauser

Nameplate . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Network configuration . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
Number of devices. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35

O

Onsite display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Operating elements, function . . . . . . . . . . . . . . . . . . 32–33
Operating elements, position. . . . . . . . . . . . . . . . . . . . . . . 31
Operational safety. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Overvoltage protection . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28

P

Pipe mounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
Position adjustment, FF configuration program . . . . . . . 60
Position adjustment, FieldCare . . . . . . . . . . . . . . . . . . . . . 60
Position adjustment, onsite . . . . . . . . . . . . . . . . . . . . . . . . 32
Pressure measurement. . . . . . . . . . . . . . . . . . . . . . . . . . . . 61
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

Q

Quick Setup menu pressure (FieldCare) . . . . . . . . . . . . . . 61

R

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Repair of Ex-certified devices. . . . . . . . . . . . . . . . . . . . . . . 97
Reset . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Returning devices . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Rotating the housing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

S

Scaling OUT parameter. . . . . . . . . . . . . . . . . . . . . . . . . . . . 67
Scope of delivery . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12
Selecting the language . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Selecting the measuring mode . . . . . . . . . . . . . . . . . . . . . 58
Separate housing, assembly and mounting . . . . . . . . . . . 22
Shielding. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27
Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Software history . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 98
Spare Parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Storage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Supply voltage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27

T

Temperature isolator, installation instructions. . . . . . . . 19
Troubleshooting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81

U

Unlocking operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 52

W

Wall mounting. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
Warnings . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81

99

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Welding recommendation. . . . . . . . . . . . . . . . . . . . . . . . . 23
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

100Endress+Hauser

Cerabar S PMC71, PMP71, PMP75 with FOUNDATION Fieldbus

Endress+Hauser

10171335667

www.addresses.endress.com