Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Real-Time Digital Signal Processors withradiation detectors produced by TechnoAP
Lunch time Exhibitor presentation

Techno AP Co., Ltd.

2976-15 Mawatari, Hitachinaka-city, Ibaraki 312-0012, Japan
Phone: +81-29-350-8011, FAX: +81-29-352-9013
Web: http://www.techno-ap.com
E-mail: order@techno-ap.com
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Contents
1. Information on our new products.
2. Outline of our company (Techno AP)

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

1. New product information (DSP)
For several scintillation detectors
 APV8508-14 (8CH, 500MSPS, 14bit-ADC)
 APV8104-14 (4CH, 1GSPS, 14bit-ADC)
 APV8516-8(16CH, 500MSPS, 8bit-ADC)
 APV8702-8(2CH, 3GSPS, 8bit-ADC)
For Ge semiconductor detector, SDD, Si(Li), etc.
 APV8016(X) (16CH, 100MSPS, 14bit-ADC)* Other models are also available.
* Our products can be customized.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

1. New product information (Digitizer)
 APV7302-8(3GSPS, 2CH, 8bit, GbEther) APV7104-14 (1GSPS, 4CH, 14bit, GbEther) APV7516-8(500MSPS, 16CH, 8bit, GbEther) APV7508-14 (500MSPS, 8CH, 14bit, GbEther)* Other models are also available.
* Our products can be customized.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
TriggerPrevious methodTime

The measurement
is combination ofseveral modules.

DET

CFD

TDC

QDC

Controller(CAMAC,VME,
Energyetc.)

APV8508
Functions ofseveral modules
have been unified.

DET

19th Real-Time Conference, 2014/05/26

APV
8508
(DPP)

List-Data(Time, Energy)(Ethernet)

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
The APV8508-14 is a waveform analysis board for
scintillation detectors. Each channel (8 CH) isequipped with high-speed, high resolution ADC (500
MHz, 14 bit). The APV8508-14 is able to correspondto the high rates of more than 100 kcps per CH in thelist mode with using the Gigabit Ethernet (Gb Ether)connection.
[Functions](Digital) CFD, TDC, QDC,
(Optional) Digital PSA, Digital Coincidence[Usage Example]
The signal analysis of several high-speed
scintillation detectors.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

ADC Performance
500 MSPS 14bit (APV8508-14)
20

bitRMS = 1.3 bit(VRMS = 0.16 mV)

bitRMS =√( (Σ(xi – x )2 ) / i )xi : Outputted Datax : Average

0-20

2 ns / 1 bin

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
Analog-to-Digital Converter
1600

2 ns / point

1400
1200

digit [-]

1000
800
600
400
200
0
0

10

20

30

40

50

60

70

80

90

100

110

120

130

-200
time [ns]LaBr3 @ 662 keV (137Cs), The sampling interval is 2 ns.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
Functions;(Digital) CFD, TDC, QDC,
(Optional) Digital PSA, Digital Coincidence
Outputted the list data

TDC[55..40]
TDC[39..24]
TDC[23..8]
TDC[7..0], TDCFP[7..0]CH[2..0], QDC[12..0]

These data is transferred to the PC via Ethernet.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

I’ll explain about the APV8508 functions;
CFD and the TDC.
TriggerPrevious methodTime

The measurement
is combination ofseveral modules.

DET

CFD

TDC

QDC

Controller(CAMAC,VME,
Energyetc.)

APV8508
Functions ofseveral modules
have been unified.

DET

19th Real-Time Conference, 2014/05/26

APV
8508
(DPP)

List-Data(Time, Energy)(Ethernet)

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
CFD and TDC

List Format
TDC[55..40]

Time stamp timing by CFD waveform
1600

TDC[39..24]
TDC[23..8]

Delay

2ns / point

1400

TDC[7..0], TDCFP[7..0]

1200

CH[2..0], QDC[12..0]

digit [-]

1000
800
600

out
(t) = In (t-delay) + (-f ) ・ In (t)
1600

400
200
0
-200

0

10

20

30

40

50

60

70

80

90

100

110

120

1400

130

time [ns]1200＋

TDC Timing

1000

digit [-]

800

1600
1400
1200

600
400

1000

digit [-]CFD WALK

200

800
600

0

Invert of Function times

400
200

0

10

20

30

40

50

60

70

80

90

100

110

120

130

-200

0
-200

0

10

20

30

40

50

60

70

80

90

100

110

120

130

-400
time [ns]-400time [ns]19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
CFD and TDC
Time stamp timing by CFD waveform
1600

delay

2ns /point

1400

Delay ： 2ns, 4ns, 6ns, 8ns, 10ns, 16ns, 22ns, 28nsout (t) = In (t-delay) + (-f ) ・ In (t)Function ： 0.03 ～ 0.48 (interval
0.03)
CFD WALK ： any value
1200

digit [-]

1000
800
600
400

1600

200

1400

0

-200

0

10

20

30

40

50

60

70

80

90

100

110

120

130

1200

time [ns]

TDC Timing

1000

＋

digit [-]

800

These parameter setting is very easy by the PC.
1600
1400

600
400

1200

CFD WALK

200

1000

digit [-]

800
600

0

Invert of Function times

400

0

10

20

30

40

50

60

70

80

90

100

110

120

130

-200

200

-400

0
-200

0

10

20

30

40

50

60

70

80

90

100

110

120

130

time [ns]-400time [ns]19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

digit [-Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

CFD and TDC

600 2ns / point

It can calculate the time stamp in less than 2 ns.
1600
1400
1200

List Format
TDC[55..40]
TDC[39..24]

1000

digit [-]

TDC[23..8]

400

800
600
400

TDC[7..0], TDCFP [7..0]CH[2..0], QDC[12..0]

200
0
0

10

20

30

40

50

60

70

80

90

100

110

-400

120

130

Resolution 2 ns / 8 bit-200

19.297 ns

200
time [ns]

CFD WALK

00
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

I’ll explain about the APV8508 function;
QDC.
TriggerPrevious methodTime

The measurement
is combination ofseveral modules.

DET

CFD

TDC

QDC

Controller(CAMAC,VME,
Energyetc.)

APV8508
Functions ofseveral modules
have been unified.

DET

19th Real-Time Conference, 2014/05/26

APV
8508
(DPP)

List-Data(Time, Energy)(Ethernet)

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

QDC (Charge to Digital Converter)

List Format
TDC[55..40]

QDC ： Integral Mode
1600

2 ns / point

1400

digit [-]

TDC[39..24]
TDC[23..8]

1200

TDC[7..0], TDCFP[7..0]

1000

CH[2..0], QDC[12..0]

800
600

QDC = Σfilterout(t※)※ t = integra range

1600

400

1400

200

8 ns / point

0
0

10

20

30

40

50

60

70

80

90

100

-200

110

120

1200

130

time [ns]digit [-]

1000 Filter

Setting;
Pretrigger sel ：-8ns, -16ns, -24ns, -32ns, -40ns
Integral range ：0ns ～ 32 μs (8ns interval)
Filter ：Ext, 8ns, 16ns, 24ns, 32ns, 48ns, 64ns
19th Real-Time Conference, 2014/05/26

800

Integral
range

600
400
200
0
0

Pretrigger select

38

78

118

time [ns]Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

List Format

QDC (Charge to Digital Converter)

TDC[55..40]

QDC ： Peak Mode

TDC[39..24]
TDC[23..8]
TDC[7..0], TDCFP[7..0]CH[2..0], QDC[12..0]

QDC = PeakData
1600

2 ns / point

Peak Data

1400
1200

digit [-]

1000
800
600
400
200
0
0

10

20

30

40

50

60

70

80

90

100

110

120

130

-200
time [ns]19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

PSA (Pulse Shape Analysis)

List Format
PSA FAST[12..0]※option
PSA SLOW[12..0]※option
TDC[55..40]

1600

peak

TDC[39..24]

Wait cnt from CFD-THR-Discri

TDC[23..8]

1400

TDC[7..0], TDCFP[7..0]
1200

CH[2..0], QDC[12..0]

digit [-]

1000

FAST

800
600

SLOW

400
200
0
0

10

20

30

40

50

60

70

80

90

100

110

120

130

-200
time [ns]19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
Summary 1
1.The APV8508 is a multichannel, 8 CH.
2.The APV8508 has the multifunction, such as CFD,
TDC, QDC etc.
3.The APV8508 can also measure a high-count rate.
4.The APV8508 is using the Gigabit Ethernet(GbEther).

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
Summary 2
5.The APV8508 can display the waveform on the PC.
Therefore, the user is able to set very easily, such as
the threshold and WALK etc.
6.The APV8508 can save a waveform. Therefore, theuser can analyze the data.
7.The APV8508 can hold a time information at a longtime in the LIST mode.
(This was impossible to do with the previous TAC
and TDC functions.)
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8104-14 (4CH, 1GSPS, 14bit-ADC)
The APV8104-14 is a waveform analysis board.
Each channel (4 CH) is equipped with ADC (1GHz, 14 bit). The characteristic of this board isusing a 1 GHz, 14 bit ADC. The APV8104 is able tocorrespond to the high rates of more than 100 kcps
per CH in the list mode with using the Gigabit
Ethernet (Gb Ether) connection.

Functions;(Digital) CFD, TDC, QDC,
(Optional) Digital PSA, Digital Coincidence
Usage Example：
The signal analysis of several high-speed
scintillation detectors.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8516-8 (16CH, 500MSPS, 8bit-ADC)
The APV8516-8 is a waveform analysis board. The
characteristic of this board is 16 CH. Each channel(16 CH) is equipped with ADC (500 MHz, 8 bit).
The APV8516-8 is able to correspond to the highrates of more than 100 kcps per CH in the list modewith using the Gigabit Ethernet (Gb Ether)connection.

Functions;(Digital) CFD, TDC, QDC,
(Optional) Digital PSA, Digital Coincidence
Usage Example：
The signal analysis of several high-speed
scintillation detectors.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8702-8 (2CH, 3GSPS, 8bit-ADC)
The APV8702-8 is a waveform analysis board. The
characteristic of this board is using 3 GSPS, 8 bitADC. The APV8702-8 is able to correspond to thehigh rates of more than 100 kcps per CH in the list
mode with using the Gigabit Ethernet (Gb Ether)connection.

Functions;(Digital) CFD, TDC, QDC,
(Optional) Digital PSA, Digital Coincidence
Usage Example：
The signal analysis of several high-speed
scintillation detectors.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

ADC Performance (APV8104)
1 GSPS 14bit
20

bitRMS = 1.5 bit(VRMS = 0.18 mV)

bitRMS =√( Σ(xi – x ) 2 / i )xi : Outputted Datax : Average

0-20

1 ns / 1 bin

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

3.8

1200

3.6

1000

Throughput[kcps]

Energy Resolution[%]Energy Resolution and Throughput (APV8104)

3.4
3.2
3
2.8

800
600
400

2.6

200

2.4

0
0

200

400
600
Throughput[kcps]

800

1000

Throughput vs Energy resolution

0

300

600
Input[kcps]

900

1200

Input vs Throughput

Detector: LaBr3 scintillation detector
Radiation source: 137Cs (10 MBq), 60Co (2 MBq)
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Usage Examples

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)

Detector: LaBr3 scintillation detector
Radiation source: 22Na @ 511 keV
Mode: Wave mode

Inputted waveform can confirm on the PC.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
Yellow: Origin waveform
Blue: CFD waveform shaping

Detector: GSO scintillation detector
Radiation source: 22Na @ 511 keV
Mode: Wave mode

It can measure early rise time and characteristic fall time like the LaBr3 detector.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
Detector: LaBr3 scintillation detector
Radiation source: 22Na @ 511 keV
Mode: Wave mode, CFD waveform shaping

You can check the CFD waveform shaping.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
List data
TDC[55..0]

Inputted TDC of 2 CH is used to the measurementof time resolution.
22Na @ 511 keV

FWHM = 230ps
TDCFP[7..0]

LaBr3(Ce) scintillation detector
vs LaBr3(Ce) scintillation detector
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
22Na @ 511 keV

FWHM / FWTM
= 2.7 ns / 5.7 ns

The time resolution of the GSO scintillation detector
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
An energy spectrum of LaBr3(Ce) scintillation detector.
The histogram used the QDC (Integral Mode) list data.
List data
QDC[12..0]

Real-time

137Cs energy resolution: 2.9 % @ 662 keV
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8508-14 (8CH, 500MHz, 14bit-ADC)
An energy spectrum of GSO scintillation detector.

22Na energy resolution: 13 % @ 511 keV
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8702 (2CH, 3GHz, 8bit-ADC)

FWHM = 192ps

Waveform Mode

Time resolution Mode

(BaF2 scintillation detector, 22Na @ 511 keV)
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8702 (2CH, 3GHz, 8bit-ADC)

FWHM = 495ps

Waveform Mode

Time resolution Mode

(LYSO scintillation detector, 22Na @ 511 keV)
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

APV8016(X) (16CH, 100MSPS, 14bit-ADC)
The APV8016 is a digital signal processor for
gamma-ray. The APV8016 can input a direct preamp signal from the Ge semiconductor detector.
The inputted signal is converted to digital signal
processing by high-speed ADC (100 MHz, 14bit)
and highly-integrated FPGA. The measurement
data is a histogram, an event and a waveform. That
data is transferred to the PC via Ethernet.
※ The APV8016X model is used for X-ray. The

APV8016X can input a direct pre-amp signal fromthe detector, such as SDD, Si(Li), SSD, SiPin etc.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

DSP (Digital Signal Processor)
Our DSP is a Multi Channel Analyzer (MCA)
equipped with a real-time digital signal processing
function.
DSP (Digital Signal Processing)

DetectorPC
Pre AMPFPGA
Diｆｆerential

AMP

Coarse gain
Fine gain

ADC

100MSPS
14bit

Digital Pulse
Shaping

Timing
Logic

FIFO

CPU &
Ethernet

LAN
Histogram
MemoryOscilloscope

DAC

The block diagram of the DSP circuit board
The data collect mode is a histogram mode, an event mode, and a wave
mode.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Characteristic test of DSP using HPGe detector
Detector

PGT Coaxial P-type HPGe size:10% nominal energy resolution:1.8keV@1.33MeVHigh Voltage+2000V

Source60Co

Pulse Shaping(1)DSP Trapezoidal Filter(2)Analog Gaussian Filter

MCA

APV8016 internal MCA

TechnoAP AnalogMCA typ.APV7400

FWHM[keV]60

Co@1.33MeV

2.5

Analog2

DSP
1.5

DSP:TrapezoidalFilter(risetime=6us)1

Analog:Gaussian Filter(shapingtime=3us)
0.5
0

20

40

60

80

100

ICR[kcps]

Characteristic of energy resolution
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Characteristic test of DSP using HPGe detector
80
DSP:TrapezoidalFilter(risetime=6us)
Analog Gaussian Filter(shapingtime=3us)

70

DSP

60

OCR[kcps]

50
40
30

Analog

20
10
0
0

20

40

60

80

100

120

140

160

180

ICR[kcps]
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

DSP resolution 1.7 keV @ 60Co, 1.33MeV

Energy spectrum (detector: HPGe detector, radiation source: 60Co, count rate: 2 kcps,
shaping time: 6 micro seconds, measurement time: 15 min, detection efficiency: 10 %)19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

With the high count rate the energy resolution is 1.76 keV.

Energy spectrum (detector: HPGe detector, radiation source: 60Co, count rate: 20 kcps,
measurement time: 15 min, detection efficiency: 10 %)19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Usage Example １: Ge semiconductor Compton camera
Stack
Strip Ge
SamplePreamp output
13 CH
13 CH
13 CH
13 CH

GREI (Gamma-Ray
Emission Imaging)

List(Time, PHA,
Calculation…)

PC

Composition Example
Application for APV8013A

External
From RIKEN Next-generation Imaging Team.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Usage Example 2: Coincidence Doppler Broadening
Measurement of Positron Annihilation (CDB)<Composition Example>
Ge#1

It is able to support the coincidence processing
using several detectors.

511 keV
22Na

PC

511 keVGe#2
In the position annihilation
field, the Coincidence
Doppler Broadening
Measurement of Positron
Annihilation for thedetection of micro-void of
the materials, such as
semiconductor etc.
19th Real-Time Conference, 2014/05/26<Coincidence 2D-Map>

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Digitizers• APV7302-8(2CH, 3GSPS, 8bit)• APV7104-14 (4CH, 1GSPS, 14bit)• APV7104-12 (4CH, 1GSPS, 12bit)• APV7108-8 (8CH, 1GSPS, 8bit)• APV7508-14 (8CH, 500MSPS, 14bit)• APV7508-12 (8CH, 500MSPS, 12bit)• APV7516-8 (16CH, 500MSPS, 8bit)• APV7016-14 (16CH, 100MSPS, 14bit)
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

At this time, we have showed all of our products in
VME-type.
You can choose between the VME-type or the Unittype because our products can be customized.
Please contact us if you need further information.

Contact us: order@techno-ap.com

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

Our other products:
 Gamma Imaging Module Spectrometers
 MCA (Multi Channel Analyzer)
 High-Voltage Power Supply Power Supply for Preamp Preamps
 Detectors (LaBr3, BaF2, GSO, LFS, etc)
 NIM module

For more information, please visit our booth.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

2. Techno AP outline
Techno AP is specialized in development, manufacturing
and sales in the radiation measurement field. All of our
products are developed by our company.
We are also doing the commissioned development fromthe research facilities and the universities.

Main clients: Major companies,
Public research organizations,
Universities.

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

2. Techno AP outline
Techno AP

19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

2. Techno AP outline[Business lineup](1) Sale of radiation measuring instrument, and radiation counter.
(2) Development of radiation measuring device, and radiation counter.
(3) Development of research and development device, measurement
controlling system, and inspection apparatus.
[Contact us]
Address; 2976-15 Mawatari, Hitachinaka-city, Ibaraki 312-0012,
JAPAN
Phone; +81-29-350-8011
FAX;+81-29-352-9013
Web;
http://www.techno-ap.com/index_e.html

E-mail; order@techno-ap.com
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.

Title, Contents, New products lists, DSP, Digitizer, Products information, Company outline

2. Techno AP outline

Thank you for your timeand for your attention.

The End.
19th Real-Time Conference, 2014/05/26

Techno AP Co., Ltd.