Battery BAL40
Data sheet

LINAK.COM/MEDLINE-CARELINEBAL40
The BAL40 lead acid battery pack is part of the LIFT40
product series specially developed for patient lifts.
LIFT40 is a complete system consisting of the control box
CAL40 or CAL40+, a battery BAL40 and an internal charger
or external charging set comprising the CHL40 cradle andthe wall plug charger CH01 for mounting on the wall or
positioning on any flat surface. Combined with one or moreactuators and a hand control you have a complete system formodern patient lifts and sit-to-stand lifts.
The BAL40 is a next-generation lead acid battery. Housedin an ergonomic package for fast and single-handed battery
exchange.

Features and options:• Capacity: 2.9 Ah• Weight: 2.6 kg• Cycles per charge (lifts/lowerings): Approx. 40 cycles*• Charging time: With internal charger approx. 4-5 hours withexternal charger CH01 approx.5-6 hours• Chemistry: Lead acid
• Voltage: 24 V
• IP rating: IPX4 or IPX6
• Colour: Signal White, RAL 9003
* Number of lifts and lowerings with a fully charged battery
under the following conditions: Normal speed (CAL40 and LA44),
thrust up to 6,000 N
Stroke: S = 200 mm. The actuator does not reach the end
position at any time.

LINAK.COM/MEDLINE-CARELINE

Usage:• Compatibility: CAL40/CAL40+/CHL40• Duty cycle: Max. 10% or 2 minutes continuous use followed by
18 minutes without use at a max. discharge current of 10 A
• Charging: Via external wall charger CHL40 and CH01 or via
CAL40 or CAL40+ with internal charger.
• Recharging during storage: First battery recharge must be nolater than 6 months after the production date stated on thelabel. Hereafter, the battery must be recharged at least every
6 months.
• Operating temperature: +5 °C to +40 °C• Charging temperature: +5 °C to +40 °C Charging at highambient temperatures can impact the charging time.
• Storage temperature: -15 °C to +40 °C(+10 °C to +25 °C - recommended) The batteries must be
stored in an applicable storage room without direct sunlight.
• Relative humidity: 20%-80% – non-condensing• Atmospheric pressure: 700 to 1060 hPa• Meters above sea level: Max. 3000 meters• Approvals: IEC 60601-1
ANSI/AAMI ES60601-1
CSA CAN/CSA-C22.2 NO. 60601-1
IEC 60601-2

Ordering example
BAL40
BA L40 D 4 A 29 0 0 000
Reserved for future use
Software

0 = No software

Reserved for future use
Capacity

2.9 AhTechnologyA = Lead acid

IP degree

4 = IPX4
6 = IPX6Colour

D = Signal White RAL 9003

Model

L40 = LIFT™ 40 series

FamilyBA = Battery Box

Dimension drawing

Drawing no.: 1048W4000

Copyright® LINAK 2020.12 MA M9-02-882-A . Chapter 12.3
LINAK® accepts no responsibility for possible errors or inaccuracies in catalogues, brochures, and other material. LINAK reserves theright to change its products without prior notice. LINAK cannot guarantee product availability and reserves the right to discontinue
the sale of any product. The user is responsible for determining the suitability of LINAK products for a specific application. All sales aresubject to the ‘Standard Terms of Sale and Delivery’, available on LINAK websites.
LINAK and the LINAK logotype are registered trademarks of LINAK A/S. All rights reserved.

FOR MOUNTING INSTRUCTIONS AND GUIDANCE IN USAGE, PLEASE SEE THE RELEVANT USER MANUAL