Power Factor Meter (Transducer Type) - PTF
POWER FACTOR METER (1)TypeRating (2) (3)PF-17-□-1215-□-1210-□-12PBF-17-□-3315-□-3310-□-33PF-17-□-3315-□-3310-□-33
PBF-17-□-3415-□-3410-□-34
PF-17-□-3415-□-3410-□-34

Application
Single phase
3 phase(balanced)
3 phase(unbalanced) (4)3 phase 4-wire(balanced)
3 phase 4-wire(unbalanced) (4)Consumption VA

Accessory(Transducer)

Voltage side

Current side

110V, 5A(1A)

1VA

1VA

220V, 5A(1A)

2VA

2VA

110V, 5A(1A)Each phase 1VA

Each phase 1VA

220V, 5A(1A)

Each phase 2VA

Each phase 2VA

110V, 5A(1A)

Each phase 1VA

Each phase 1VA

220V, 5A(1A)

Each phase 1VA

Each phase 2VA

110V, 5A(1A)

Each phase 1VA

Each phase 1VA

220V, 5A(1A)

Each phase 2VA

Each phase 2VA

110V, 5A(1A)

Each phase 1VA

Each phase 1VA

220V, 5A(1A)

Each phase 2VA

Each phase 2VA

PT-62M-12
PBT-62M-33
PT-63M-33 (5)PBT-62M-34
PT-64M-34 (5)

Note:
(1)
Standard Scale: Lead0.5～1～Lag0.5. Scale for 3 phase 3-wire balance only: Lead0～1～Lag0.
(Effective measuring range: Lead0.3～1～Lag0.3 also can be manufacture)
Please speficy frequency (50Hz or 60Hz) all type except 3 phase balanced circuit.
(2)
When above rating is exceeds, please external CT or VT respectively to meter 110V, 5A(1A).
Usable voltage range: 110V: 90～130V; 220V: 180～260V
Please use in positive phase sequence.
(3)

Guaranteed allowance is over than 1/5 of the rated current.

(4)

3 phase (unbalanced), 3 phase 4-wire (unbalanced) is voltage balanced.

(5)

Refer to next page for the connection diagram.

Dimension

Dimesion for Type: PF-17, 15

Dimension for Type: PF-10* n = unit of meter

TypeABC

DE

PF-17

168

170

4163

PF-15

148

150

0145

F
1 unit2-unit or above

164±0.5

41 +1-0(41×n)+1-0

146±0.5

41 +1-0(41×n)+1-0

Please assemble the meter in the center of the panel if you need continuous assemble.
Or please contact and discuss with us if you need to continuous assemble more than 10 units.

TypeWeight (g)1 pointer2-pointer

PF-17Below 1200

Below 2100

PF-15

Below 1200

Below 2100

PF-10

Below 960

18Power Factor Meter (Transducer Type) - PTF
Connection Diagram (If make a mistake on phase sequence, it becomes error)

Single Phase Power Factor Meter
External with PT-62M-12

3 Phase 4 wire (balanced)
Power Factor Meter
External with PBT-62M-34

3 Phase Balanced Power Factor Meter
External with PBT-62M-33

3 Phase unbalanced Power Factor Meter
External with PT-63M-33

3 Phase 4 wire unbalanced
Power Factor Meter
External with PT-64M-34

Dimension

P(B)T-62M-□

PT-63M-33

PT-64M-34

19Frequency Meter (Transducer Type) - AF
Rated Voltage

110V

Measurement Range
45~55Hz55~65Hz45~65Hz(1)
350~450Hz
45~55Hz55~65Hz45~65Hz350~450Hz (1)(2)

220V (2)Consumption VAAccessory

AF-17, 15, 10(Transducer)

1.7VA

FT-62M

90~130V

2.5VA

FT-62M

180~260V

Voltage Change Range

Note:
(1)
Special frequency measurement range also can be manufacture (up until 1000Hz)(2)
Usable voltage range: 110V: 90～130V; 220V: 180～260V
Please contact with us for manufacture the above rated voltage and voltage change range.
For SCR Wareform MeterMeter SCR waveform input (Distortion waveform) also can be manufacture.
Type name: AF-□H-□-□

Connection Diagram

Dimensions

Frequency meter
external with FT-62M

Dimensions for Type: FT-62M

Dimensions

Dimesion for Type: AF-17, 15

Dimension for Type: AF-10* n = unit of meter

TypeABC

DEF
1 unit2-unit or above(41×n)+1-0(41×n)

+1-0

AF-17

168

170

4

163

164±0.5

41 +1-0AF-15

148

150

0

145

146±0.5

41 +1-0Please assemble the meter in the center of the panel if you need continuous assemble.
Or please contact and discuss with us if you need to continuous assemble more than 10 units.

Type

Weight (g)1 pointer2-pointer

AF-17

800

1100

AF-15

800

1100

AF-1056020