SQLC-215-133

INSTRUCTION MANUALPOWER LINE SUPER MULTI-METERSQLC-110L
3-PHASE 3-WIRE
VOLTAGE, CURRENT INPUT 2VT･3CT
CURRENT INPUT 3CT
ANALOG OUTPUT
HARDWARE MODEL G [White backlight]

SQLC-215-133

Introduction
Thank you for your purchase of our product.
Read this instruction manual carefully before installation, wiring, and using this product.
Keep this instruction manual handy for reference at any time.
Have a contact with us or sales agent in case that this instruction manual is lost or damaged.
<Caution>
Have a contact with us when you have any questions or are aware of missing article.
Safety precaution
Important contents are mentioned in this instruction manual to prevent any damage / use this product appropriately.
Keep the following safety precaution in mind after understanding each sign.
Improper use may lead to death or severe injury.
Improper use may possibly lead to death or severe injury.
Improper use may lead to medium injury.
 We are not responsible for the damage caused by following condition (earthquake / fire which is not caused byus, action by third party, other accident, damage caused by our customer, misuse, product usage under abnormal
condition).
 We are not responsible for secondary damage caused by product use / product malfunction(loss of profit, haltof business operation). We are also not responsible for damage caused by false operation in combination withconnecting equipment which is beyond our control.

 Do not disassemble, remodel and repair this product.
Have a contact with us or sales agent when product failure happens to prevent fire / electric shock / injury.
 Do not get this product wet to prevent heat generation / ignition / product failure. When this product gets
wet, stop using it.
 Do not connect metal excepting wiring to terminal in order to prevent heat generation/ignition.
 Do not get this product near the inflammables / combustible chemicals / gas to prevent fire.

 Connect specified power supply.
Connecting power supply beyond specification causes fire / product failure.
 When dust is on the terminal, wipe it off after power is OFF to prevent fire.
 Follow the below-mentioned procedure when abnormality (fuming / bad odor) happens.
(1) Stop a power supply and input, and stop using.
(2) Contact our company or a distributing agent.

 Do not use this product in a environment of high temperature / high humidity to prevent any damage.
 Do not touch the terminal during operation to prevent electric shock.
 Do not pull/bend connecting cable with force. Cable damage causes heat generation / burn and contact failure
leads to equipment damage.
 Do not connect/inspect with wet hands to prevent electric shock.

1

SQLC-215-133

Other precaution
 Don't mount or store this unit in the following environment.
Places where corrosive gas (SO2 / H2S / etc.)(１) is generated.
Places where dust is generated.
Places with much vibration and shock.
Places with influence of external magnetic field (２).
Note(１) Corrosive gas = Sulfur dioxide SO2 / Hydrogen sulfide H2S / etc.
Note(２) Large current bus / saturable reactor / etc.
 Wipe off dirt on the surface with dry cloth softly. Keep in mind that strong rubbing of nameplate leads tocharacter disappearance. Organic solvent and chemicals and cleaner is not appropriate for cleaning.
 Mercury component, Nickel-cadmium battery are not used in this product.
 Please dispose of this product as industrial waste (noncombustible).
 The precautions at the case of using by outdoor board.
① This product is not dust-proof structure and not waterproof structure. In using by the outdoor panel,please avoid the place, which dust causes. And, please install in the place that requires neither rain
nor waterdrop.
② Please do not install in the place of sunlight. A display of this product may become hard to see.
And, deformation of a case may take place by the surface temperature rise.
The warranty period and warranty scope.
 Warranty period.
The warranty period of this product is for one year after supplying the appointed place.
 Warranty scope.
In the state of the normal use of product-specification within the range according to this instruction manual,
the trouble within the warranty period performs exchange or repair gratuitously.
However, the shipping expenses and the packing cost in the case of shipping obtain as payment on a customer.
And, if it corresponds to the next, it does not warrant.
(1) If it breaks down when converted or repaired except our company.
(2) If it breaks down by use out of specification range.
(3) If the cause of trouble is based on cause other than this product.
(4) Transportation, movement, damage by falling, and trouble.
(5) A natural disaster, disaster, etc., if it is the trouble which is irresponsible for a payment side(our company or distributing agent).
This warranty is a warranty of a product simplex. It cannot warrant the damage induced by trouble of this product.
Change of instruction manual written contents.
This instruction manual changes written contents without a notice by product improvement etc.

2

SQLC-215-133

The precaution for use.
1. Environment of usage and storage.
Don’t mount or store this product in the following environments. If the unit becomes defective due to the use in
an environment other than specified, it may be repaired for pay even during its warranty period (one year after thedate of delivery).
① Don’t mount or store the unit at a place where the ambient temperature is other than a range of -10 to +55℃ or
the relative humidity is higher than 85%RH.
② Don’t mount or store the unit at a place where a corrosive gas such as SO2, H2S, etc. is generated.
Corrosive gas = Sulfur dioxide SO2 / Hydrogen sulfide H2S / etc.
Large current bus / saturable reactor / etc.
③ Don’t mount or store the unit at a dusty place.
④ Don’t mount or store the unit at a place subjected to noticeable vibrations or shocks.
⑤ Don’t mount or store the unit at a place subjected to noticeable external noises.
⑥ Don’t mount or store the unit at a place exceeding the altitude of 2000m.
⑦ If this unit directly measures an inverter output of cycle control, SCR phase angle control or PWM control, anerror may increase due to its operation principle.
2. Cautions on use in an outdoor panel
Be careful with the following items when using this unit in an outdoor panel.
① Don’t mount the unit at a place where it is directly exposed to rain or water drops, otherwise this unit may
become defective because of no water-proof or drip-proof structure.
② Don’t mount the unit at a dusty place.
③ Don’t mount the unit at a place exposed to direct sunlight. Avoid exposing the unit to direct sunlight eventhrough a glass window.
If the meter is directly exposed to sunlight, the surface temperature of the meter rises and the case may be
deformed if the temperature exceeds 80℃.
④ If the average temperature around the meter exceeds 40℃, the life of the unit may shorten.
3. About dew condensation
If the temperature and humidity of an installation change rapidly when a product is a non-energization, thewaterdrop by dew condensation may adhere to a display inner side. (The display filter and the LCD surface stickand the pattern of the shape of a circle or an ellipse occur.)This phenomenon improves by doing the power distribution of the auxiliary power and leaving it for about 2 hours.
4. Mounting and wiring
Mount and connect the unit by a technician while referring to the instruction manual and observing the following
cautions.
 Connect the unit after confirming the connection diagram. An improper connection may cause troubles
such as the generation of a high voltage on the secondary side of its current transformer or burningof the unit or the occurrence of a fire.
 Don’t perform any connection in a hot line without turning off the power supply in advance, otherwise
an electric shock accident, troubles or burning of the unit, a fire, gas explosion, or other very
dangerous accidents may occur.
 The terminal cover is mounted for preventing an electric shock accident. Mount the terminal cover
without fail after the end of work.
5. SettingThis unit requires setting and confirmation of the measuring range, etc. before use. Wrong setting, if any, causes
malfunction of the unit. If setting should be wrong, neither measurement nor output becomes correct.
Carefully read the instruction manual before setting the unit.
■ Default setting
The initial setting values are as specified below at the delivery time. Set them according to the working conditions.
The unit will be delivered with your specified setting values, if so specified.

3

SQLC-215-133① Voltage, current input (1/2)№

Setting item

1

Display combination

2

Alarm output(３)

3

Demand detection

4

Harmonic detection

5

Instant measurement
detection

6

Backlight

Pattern
Main monitor
Sub-monitor (Left)
Sub-monitor (Center)
Sub-monitor (Right)
Bar graph
FactorReset form
Alarm 1
Contact delay time
Test
FactorReset form
Alarm 2
Contact delay time
Test
Upper limit value
Demandcurrent
Interval
Upper limit value
Interval
Demandactive power Operation form
Power-factor operation form
Distortion-factor upper limit
5th conversion content rate
upper limit
Current
Factorn-th content rate
Upper limit
Distortion-factor upper limit
5th conversion content rate
upper limit
Voltage
Factorn-th content rate
Upper limit
5th conversion detection characteristics
Average value time limit
Voltage upper limit value
Voltage lower limit value
ActionBrightness

3-phase 3-wire
110V input
220V input
Pattern 1
A(L2)
V(L1L2)
W
Wh
A(L2)
DA
AUTO
0 second－DA
AUTO
0 second－
80.0A
0 secondOFF
0 second
Operating system according with bimetallic
type.
Instant measurement
OFF
OFF
5th
OFF
OFF
OFF
5th
OFF
Inverse-time-delay mode
0 minute
OFF
OFF
AUTO
3（Middle）

Voltage range
6600V
Current range
100.0A
Current display peculiar sensitivity
100.0A
Active power polarity
One-side swing
7 Measurement range
Active power range
1200kW
Reactive power range
600kvarPower-factor range
LEAD 0.5‐1‐LAG0.5
Frequency range
45‐65Hz
Output factor 1
A(L2)
Output factor 2
V(L1L2)
Output factor 3
W
Output factor 4cosφ
Analog output
8(３)
Current
100.0％
Output peculiar
Active power
100.0％
sensitivity
Reactive power
50.0％
Low input cut
OFFFactorWh
Output 1
Pulse unit
10kWh/pPulse output
9(３)
FactorWh
Output 2
Pulse unit
10kWh/p３
Note( ) A setting item is not displayed in case there is no corresponding option.

4

220V

40.0kW
20.00kvar

0.1kWh/p
0.1kWh/p

SQLC-215-133① Voltage, current input (2/2)
№
10

11

12
13
14

15

Setting item
Input 1 function
Input 2 function
Voltage
Current
Active power
Reactive power
Power-factor
Measurement display Frequency
ON/OFF
Watt-hour of power receiving
Watt-hour of power transmission
var-hour of power receiving
var-hour of power transmission
Harmonic current
Harmonic voltage
Phase line change
Input circuit
Input voltage
Dead band
Measurement
Tidal current measurement
Analog output specification(４)(５)
Bias adjustment
Output 1
Span adjustment
Bias adjustment
Output 2
Analog output
Span adjustment
adjustment
Bias adjustment(４)
Output 3
Span adjustment
Bias adjustment
Output 4
Span adjustment
External operation
input (４)3-phase 3-wire
110V input
220V input
Alarm resetMax./Min. reset
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
3φ3W (Fixation)
110V
220V
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

Note(４) A setting item is not displayed in case there is no corresponding option.
The external operation input constitutes initial value with an alarm-output option.
It becomes the next function in case there is no alarm-output option. Input 1 function：Max. / Min. reset,
Input 2 function：Measurement factor change.
Note(５) A setting item is not displayed if analog output is except DC0‐5V (or DC1‐5V) specification.

5

SQLC-215-133② Current input№1

Display combination

2

Alarm output(６)

3

Demand detection

4

Harmonic detection

5

Backlight

Setting item
Pattern
Main monitor
Sub-monitor (Left)
Sub-monitor (Center)
Sub-monitor (Right)
Bar graph
FactorReset form
Alarm 1
Contact delay time
Test
FactorReset form
Alarm 2
Contact delay time
Test
Upper limit value
Demandcurrent
Interval
Distortion-factor upper limit
5th conversion content rate
upper limit
Current
Factorn-th content
rate
Upper limit
5th conversion detection characteristics
Average value time limit

3-phase 3-wire

5thOFF
Inverse-time-delay mode
0 minute

ActionBrightness

AUTO
3（Middle）

Pattern 15
A(L2)
A(L1)
A(L3)
－
A(L2)
DA
AUTO
0 second－DA
AUTO
0 second－
80.0A
0 second
OFF
OFF

Current range
100.0A
Current display peculiar sensitivity
100.0A
Output factor 1
A(L2)
Output factor 2A(L1)Output factor 3A(L3)Analog output
7(６)
Output factor 4－
Output peculiar sensitivity
Current
100.0％
Low input cut
OFF
Alarm reset
External operation Input 1 function
8
input (６)Input 2 function
Max./Min. resetON
Measurement display Current
9
ON/OFF
Harmonic current
ON
10 Input circuit
Phase line change
3φ3W (Fixation)
Dead band
0.0％
11 Measurement
Tidal current measurement
General measurement
12 Analog output specification(６)(７)
1‐5V
Bias adjustment
0.0％
Output 1
Span adjustment
100.0％
Bias adjustment
0.0％
Output 2
Analog output
Span adjustment
100.0％
13 adjustment
Biasadjustment
0.0％(６)Output 3
Span adjustment
100.0％
Bias adjustment
0.0％
Output 4
Span adjustment
100.0％
６
Note( ) A setting item is not displayed in case there is no corresponding option.
The external operation input constitutes initial value with an alarm-output option.
It becomes the next function in case there is no alarm-output option.
Input 1 function：Max. / Min. reset, Input 2 function：Measurement factor change.
Note(７) A setting item is not displayed if analog output is except DC0‐5V (or DC1‐5V) specification.
6

Measurement range

6

SQLC-215-133

6. Operation
Be careful with the following cautions during use.
 Use the input within the rated range. Be careful since negligence of this caution may cause troubles of the unit.
 There is a function to hold the maximum value and the minimum value with a measurement factor in this product.
A blackout is guaranteed and this value isn't also cleared by a power supply reset. However, the minimum value
may be updated in case input is not applied to a power up. For this reason, in order to make the past minimum
value hold by powering on, please apply input within 1 second after switching on a power supply.
 The maximum value, a minimum value measurement factor
Measurement factor

Maximum value
measurement

Minimum value
measurement

○

○

○

×

Voltage，Current，Demand current，Active power，Demand active power，Reactive power，Power factor，Frequency
Harmonic (Distortion factor ，Effective value ，Content)

 Be careful not to touch any terminal when power is applied to the unit.
 Don’t disassemble or modify this unit without any previous permission of our company, otherwise thewarranty does not apply to the unit any more. Also, modifications may cause troubles, a fire, orother accidents. For specifications change, etc., please contact us.
7. Maintenance and check① Please wipe a surface with the dry soft cloth. A display may be lighted if a LCD face is wiped.
This is the phenomenon of occurring with static electricity.
Leave for a while, and the display goes out due to natural discharge.
Keep in mind that it will break if a LCD side is pushed strongly. And, if a filter is pushed strongly, a LCD side
may touch a filter and a pattern (a circle and ellipse) may occur. Therefore, please do not push a filter strongly.
② Check the following items.
◎ Check the unit for damage in appearance.
◎ Check if indications conform to the inputs.
◎ Check if the unit is mounted normally without any loose connections of the terminal board after turning off
the power supply.
If a question has arisen or if the unit seems to be defective, please contact us.
8. Other precautions
 Wipe off dirt on the surface with dry cloth softly. Keep in mind that strong rubbing of nameplate leads tocharacter disappearance. Organic solvent is not appropriate for cleaning.
 Neither mercury parts nor the nickel-cadmium battery is being used for this product.
 This product is disposed as an industrial waste (non-inflammable).
9. Warranty
 Warranty period
It is for one year after product delivery.
 Warranty scope
In the state of the normal use of product-specification within the range according to this instruction manual,
the trouble within the warranty period performs exchange or repair gratuitously.
However, the shipping expenses and the packing cost in the case of shipping obtain as payment on a customer.
And, if it corresponds to the next, it does not warrant.
(1) If it breaks down when converted or repaired except our company.
(2) If it breaks down by use outside the specification range.
(3) If the cause of trouble is based on cause other than this product.
(4) Transportation, movement, damage by falling, and trouble.
(5) A natural disaster, disaster, etc., if it is the trouble which is irresponsible for a payment side (ourcompany or distributing agent).
Only a product simplex is applied with a warranty. We cannot bear the responsibility such as the damage andthe loss that occurred by the use of this product and trouble or the loss of profits.
And, a warranty is effective only in Japan.

7

SQLC-215-133

Content

1. Product outline ･･･････････････････････････････････････････････････････････････ 9
1.1 Usage of product ･･･････････････････････････････････････････････････････････ 9
1.2 Features of product ････････････････････････････････････････････････････････ 9
2. The name and function of each part ････････････････････････････････････････････ 10
3. Preparation
3.1 Installation ･･･････････････････････････････････････････････････････････････ 11
3.2 Connections ････････････････････････････････････････････････････････････････ 12
4. Operation ･････････････････････････････････････････････････････････････････････ 13
4.1 The screen change and function by switch operation ･････････････････････････ 14
4.2 The kind of display ････････････････････････････････････････････････････････ 15
4.2.1 Measurement display ･････････････････････････････････････････････････････ 15
4.2.2 Alarm detection display ･････････････････････････････････････････････････ 17
4.2.3 Setting display ･････････････････････････････････････････････････････････ 19
4.3 Operation ･･････････････････････････････････････････････････････････････････ 20
4.3.1 The main monitor display-element change ･････････････････････････････････ 20
4.3.2 Phase (line) display change ･････････････････････････････････････････････ 20
4.3.3 Harmonic measurement display change ･････････････････････････････････････ 21
4.3.4 Watt-hour enlarged display ･･････････････････････････････････････････････ 21
4.3.5 Setting value check ･････････････････････････････････････････････････････ 22
4.3.6 Setting mode ････････････････････････････････････････････････････････････ 23
4.3.7 Reset ･･･････････････････････････････････････････････････････････････････ 24
5. Setting
5.1 Function table ･････････････････････････････････････････････････････････････ 27
5.2 Setting table ･･････････････････････････････････････････････････････････････ 32
5.3 Setting in detail explanation ･･････････････････････････････････････････････ 40
5.3.1 Setting mode 1 ･･････････････････････････････････････････････････････････ 40
5.3.2 Setting mode 2 ･･････････････････････････････････････････････････････････ 49
5.3.3 Setting mode 3 ･･････････････････････････････････････････････････････････ 56
6. Specification
6.1 Specification and intrinsic error ･･････････････････････････････････････････ 59
6.2 Specification, Performance ･････････････････････････････････････････････････ 62
6.3 Option ･････････････････････････････････････････････････････････････････････ 64
6.4 CE marking ･････････････････････････････････････････････････････････････････ 66
7. Maintenance and check7.1 Trouble shooting ･･･････････････････････････････････････････････････････････ 67
7.2 Check ･･････････････････････････････････････････････････････････････････････ 67
7.3 Maintenance ････････････････････････････････････････････････････････････････ 67
7.4 Test ･･･････････････････････････････････････････････････････････････････････ 68
7.5 Storage ････････････････････････････････････････････････････････････････････ 68
7.6 Countermeasures against troubles ･･･････････････････････････････････････････ 68
Appendix table 1

8

SQLC-215-133

1. Product outline
1.1 Usage of productThis single unit can measure and monitor demand-current ×3, voltage ×3, current ×3, demand active power, active-power,
reactive-power, power-factor, frequency, watt-hour, var-hour and harmonic(voltage, current).
From a low tension circuit to a high tension circuit, it is adapted for various usages, such as a measurement monitor
of a power-receiving circuit, an energy conservation power monitor, a demand current measurement monitor, and a harmonic
monitor.
The intensive monitor united with the system is made in an addendum of analog output and a pulse output.

1.2 Features of product High-performance products which collected the function of the power line multi-meter (QLC-110L, DLC-110L, HLC-110L)
of our company.
 Bar graph 1 measurement and digital 4 measurement are displayed simultaneously.
 Analog output 4 circuit and contact-output 2 circuit can be taken out. (Option)
A contact output can be selected from a pulse output, an alarm output, and a CPU error output. (Please designate
it at an order.) And, An output factor can be selected by setting.
 Two external operation inputs are possible. (Option) And, selection of reset input and a display change input ispossible at setting.
 Power supply is AC85‐264V，DC80‐143V (for both AC and DC uses).
 The mounting method of this unit is compatible with the mounting method of conventional 110 square mechanical meter.
This unit is mounted at 2 diagonal points.
 Integrated value of Wh and varh can indicate expansion to 3rd digit below the decimal point.
 Analog output is with a lower limit limiter.
 A tidal current measurement (output 2 quadrant) change is possible for var and cosφ.
 A backlight function is equipped. Selection of backlight-on, backlight-off, and auto backlight-off and settingof brightness are possible. Automatic turning off the lights at the time of non-operation can be established.
LED：White

9

SQLC-215-133

2. The name and function of each partBar graph display
The measurement value of the main monitor
is indicated by the analog.
(Setting which does bar graph displayof the measurement value of sub-monitor
is also possible.)

Scale markings
It sets automatically by measurement-range
setting.
Upper limit (or lower limit) setting index
An upper limit (or lower limit) set point isdisplayed.

SQLC-110L

Digital display
Measurement monitor can watch
4 elements at the same time.

Unit display
It sets automatically by measurement-range
setting.

･Main monitor･Sub monitor (Right)･Sub monitor (Center)･Sub monitor (Left)

Multiplying factor display
It displays on the lower right of the main
monitor at the time of electric energy and areactive energy display.

SETDISPLAY

The switch from which integrated
value of the amount of electric
power is switched to normal display(5 digits of integer) and expansion
indication (integer 2 digits ＋ below
decimal point, 3 digits) variously.
If it is not operated for 10 minutes after adisplay change, it will usually return to adisplay. It is used also as a switch whichchanges to setting mode. If it continues
pushing 3 seconds or more, it will change tosetting mode.
In setting mode, it is used as a switch that
determines a set point.

It is the switch that changes a phase(line) display of current (voltage).
If it is not operated for 10 minutes after adisplay change, it returns to the original
phase (line) display. In setting mode, it isused as a switch that terminates setting mode.

－＋

The switch to which measurement displays
element of main monitor is changed.
If it is not operated for 10 minutes after a display
change, it returns to the original measurement display
factor. In set mode, it is used as a switch that changes
a setting value.

RESET/SHIFT

MAX/MIN

It is the switch that resets various alarms.
And, in the maximum and the minimum measurement
display, it is used as a switch that resets the
maximum and the minimum value.
In setting mode, it is used as a switch to which
a setting item is shifted.

The switch to which general measurement display (usually)
and the maximum minimum measurement display are changed.
MODE

The switch to which general measurement display (usually) and theharmonic measurement (voltage and current) display are changed.
In set mode, it is used as a switch that changes a setting item.

10

SQLC-215-133

3. Preparation
3.1 Installation
Mount the unit by the attached M5 nuts to a panel of thinner than 10mm, referring to the following external dimensions
drawing and panel cutout. Fasten these nuts with tightening torque 2.75 to 3.82N･m.
● Dimension diagram● Caution on handling
Mount the LCD to obtain an optimum angle, since the contrast changes according to the monitoring angle.
For upper case installation● Installation(1) A product is put in a cut hole of a panel from a front.
A body is inserted until it exceeds retaining stopperof the lower base.
(2) Please fix a product certainly with attached M5flange nut for installation.
Please give a tightening torque as 2.75 to 3.82N･m.

11

SQLC-215-133

3.2 Connections
Please perform connection after referring to the following wiring diagram.
● Connection drawing (11)(1) 3φ3W (2VT，3CT)(2) Current input 3φ3W (3CT)

L1 L2 L3

L1 L2 L3

Note(８) Analog output, contact output, external operation input is an option.
Note(９) Can change to external reset function or external display change function by setting.
Note(10) Can choose the contact output among the pulse output, the alarm output, the CPU error output. (Designation)
・Combination of contact output
Contact output 1
Contact output 2
Pulse＋Alarm
Pulse output
Alarm output
Alarm×2
Alarm output 1
Alarm output 2
Pulse×2
Pulse output 1
Pulse output 2
Pulse＋CPU error
Pulse output
CPU error output
Alarm＋CPU error
Alarm output
CPU error output
Note(11) In case of low-voltage circuit, secondary side earthing of VT and CT is unnecessary. And, VT is unnecessary
in case it used 110V or direct 220V.

● Caution on connection(1) Mount the terminal cover without fail for safety after the end of connections.
(2) Separate the input wiring and output wiring from each other without fail, and take a preventive measure against
malfunction due to external noises.
(3) Connect the grounding terminal E (No. 7 terminal) to the ground without fail for enhancing the shield effect.
Keep the grounding resistance between the grounding terminal and the ground to be lower than 100Ω.
(4) Keep a distance of more than 30cm between this unit and the circuit breaker as well as between this unit and therelay contact signal line.
(5) No protection is necessary for this unit even if the transmission line may be affected by an induced lightning
surge or the like when transmitting an analog output to the receiver. Mount an about 500V arrester or the likebetween the line surge protector and the ground as well as between the transmission line and the ground on thereceiver side for the purpose of protecting the devices on the receiver side.
(6) It is recommended to mount a surge killer outside when connecting an inductive load to the pulse output and alarm
output. If no surge killer is mounted, the contact life may shorten.

12

SQLC-215-133

4. Operation● The function of switch

SwitchFunction

SET

The integrated value of electric energy is changed to the usual display and an enlarged display.
If it continues pushing 3 seconds or more, it will change to setting mode.
In setting mode, it is used for the determination of a set point.

RESET/SHIFT

Various kinds of alarms are reset. The maximum value and the minimum value are reset in the
maximum minimum measurement display. In setting mode, it is used for movement of a setting item.
The usual general measurement display and harmonic measurement (voltage, current) display arechanged. In setting mode, it is used for the change of a setting item.

MODE
MAX/MIN

The usual measurement display and maximum value or minimum value display are changed.

＋ ， －The measurement display element of the main monitor is changed.
In setting mode, it is used for change of a set point.

DISPLAY

A phase (between lines) display of current (voltage) is changed. It is used in case it terminatessetting mode. And, it is used in case it returns the display combination of a measurement factor.

● Convenient functions(1) In case a measurement change or a phase change is performed and the original screen composition is not clear
anymore, DISPLAY is pushed for more than 3 seconds or it's no-operation for 10 minutes and returns to original
screen structure.
(2) Even if it stops operation with setting mode, it returns to the display mode in 10 minutes.
● 7 segment displays
This product shows the guidance in various setting using 7 segment displays besides a display of a measurement value.
A digital readout and 7 segment displays corresponding to each alphabet are shown in the following.

A

B(b)

C

D(d)

E

F

G

H

I

J

K

L

M

Y(y)

Z

Non-dis Non-dis
play
play

N(n)

O(o)

P

Q(q)

R(r)

S

T(t)

U(u)

V

W

X
Non-dis
play

0

1

2

3

4

5

6

13

7

8

9

SQLC-215-133

4.1 The screen change and function by switch operation
This product changes various screens by switch operation. Here, the change step of the screen by switch operation is
explained.
Display mode
DISPLAY

Instant measurement display
Generalmeasurement
display
MODE

Current harmonicmeasurement
displaySETRESET/SHIFT

Electric-energy
integrated-value
enlarged display

Push
1 seconds

SET
or no operation,
10 minute
SET

All alarm
reset

No operation,
10 minute＋Bar graphdisplay factor
change，－No operation,
10 minute

Main monitor
display-element change

RESET/
＋ －
SHIFT

MODE

Voltage harmonicmeasurement
display

Phase (line)
display change

SET

Set point check

Bar graphdisplay factor
changeDISPLAY or
No operation,
10 seconds

MODE

SET
Push 3 seconds
MAX/MIN
Maximum and the minimum
measurement display

General measurement
maximum and minimum
display

Setting mode 1
RESET/SHIFT
Push 1 seconds

MODE

RESET/＋ ＋
SHIFT

Current harmonicmeasurement maximum
display

Push 1 seconds

Maximum / Minimum
reset(Display factor)

DISPLAY or
No operation,
10 minute
SET ＋

RESET/SHIFT

Push 3 seconds
Setting mode 2
Maximum / Minimum
reset(All factor)

MODE

DISPLAY or
No operation,
10 minute
SET ＋ DISPLAY
Push 3 seconds

Voltage harmonicmeasurement maximum
display

Setting mode 3
DISPLAY or
No operation,
10 minute

MODE

14

SQLC-215-133

4.2 The kind of display
4.2.1 Measurement display
A measurement value display has the three following types of displays.
The change of the measurement display element of the main monitor by switch operation and the change of the phase/ line display of current / voltage is possible (temporarily).
In a general measurement display, if switch operation is not performed for 10 minutes after changing a display element,
it returns to the original measurement display element automatically.
① General measurement display
Measurement factors, such as current, voltage,
and active power, are displayed.
The measurement value of four factors is
displayed at the maximum.
Setting which always displays a measurement
factor is possible. And, it is possible tochange to a display of the maximum value and theminimum value, about the measurement factor
which performs holding of the maximum value andthe minimum value by switch operation.
These maximum values and the minimum value arereset by switch operation (it updates to theinstantaneous value at the time).
In addition, as for the maximum value and theminimum value, power-supply reset is not cleared
either. And, this display is held by even after
10 minutes of switch non-operation.

Range of minimum to maximum isdisplayed in bar graph.

A(L2) display

V(L1L2)
displayWdisplayWhdisplay

General measurement display

Minimum
display

Instant
display

Maximum
display

General measurement maximum
and minimum display

・The example of a measurement display of each measurement factor (Main monitor) ･･･ Harmonic measurement is excluded.
Measurement
factorExample of display

Measurement
factorNote

Voltage

Current

Frequency

Demandcurrent

Watt-hour(Powerreceiving)

Watt-hour(Powertransmission)

Active power

Reactive power

LAG
or
LEAD
display

15

Example of display

Note“DEMAND”
is display“－” isdisplay

Demand active
power“DEMAND”
is display

Power factor

LAG
or
LEAD
displaySQLC-215-133

Measurement
factorNote

Measurement
factorvar-hour(Powerreceiving，
LAG)

“LAG”
display

var-hour(Powerreceiving，
LEAD)“LEAD”
display

var-hour(Powertransmission,
LAG)“LAG” and”－”display

var-hour(Powerransmission,
LEAD)“LEAD”and ”－”
display

Example of display

Example of display

Note② Current harmonic measurement display
It is a measurement factor display of the distortion factor of current, relative harmonic content, harmonic effective
value, etc. The measurement value of three factors is displayed at the maximum.
About sub-monitor (center) and sub-monitor (right), it is possible to change to the factor that always indicates bymeasurement. And, it is possible to change to a display of the maximum value by switch operation.
These maximum values can be reset by switch operation (it updates to the instantaneous value at the time).
In addition, as for the maximum value, even power-supply reset is not cleared. And, a display is held, even after
elapsing for 10 minutes without operating a switch.
Measurement display element)
Main monitor：Distortion factor
Sub-monitor (Left) ：5th conversion or harmonic order (n)Sub-monitor (Center)：Harmonic 5th conversion content or harmonic nth content (n=3,4,5,7,9,11,13,15)
Sub-monitor (Right) ：Fundamental-wave effective value , 5th harmonic conversion effective value or harmonic
nth effective value (n=3,4,5,7,9,11,13,15)③ Voltage harmonic measurement display
It is a measurement factor display of the distortion factor of voltage, relative harmonic content, harmonic effective
value, etc. About a function, it is the same as a current harmonic measurement display.
Distortion-factor display

Effective-value display

Distortion-factor maximum
display

Content display

Content maximum
display

Order display

Order display

Current harmonic measurement display.
Distortion-factor display

Effective-value maximum
display

Current harmonic measurement maximum display.

Effective-value display

Distortion-factor maximum
display

Content display

Content maximum
display

Order display

Order display

Voltage harmonic measurement display.

Effective-value maximum
display

Voltage harmonic measurement maximum display.
16

SQLC-215-133

4.2.2 Alarm detection display
The alarm value setting is a possible measurement factor (demand current and harmonic, etc.), it displays in casean input exceeds a set point.
Besides the usual measurement display, the detected factor is displayed on a screen upper case.
In addition, in case setting OFF (not use) as measurement factor, it does not detect.
And in the case of with an alarm-output option, it is possible to do an alarm output (relay make contact) to the outside
simultaneously with a screen display.
Alarm display possible factor) Demand current, Demand, Harmonic distortion factor (current, voltage),
Harmonic 5th conversion content (current, voltage),
Harmonic nth content (current, voltage n＝3, 4, 5, 7, 9, 11, 13, 15),
Voltage.
Alarm detection(Demand current upper limit)

A(L2) display
Alarm
detection
W display

V(L1L2) display
Wh display
General measurement display.

General measurement display (Alarm detection)

・The example of a display at the case of the detection in each alarm factor.
In case the alarm factor is indicating by measurement at the main monitor or the sub-monitor, a measurement value
constitutes a blinking display.
The displays after an alarm return.
In case a return method is automatic reset setting：It returns to the usual measurement display.
In case a return method is manual reset setting：A detection display and an alarm output hold (in case setting as an alarm output of applicable factor).
The return in this case needs alarm reset operation. Please refer to "4.3.7 Reset" about alarm reset.
Alarm factor

Demandcurrent

Upper
limit
(H)

Example of a display

Alarm factor

Detection display
Alarm setting value

Voltageharmonic
5th
conversion
content

Example of a display

Upper
limit
(H)
Alarm setting
Value

Detection
display

Setting as 5th conversion content ofbar graph (It distinguishes in anunderbar)

17

SQLC-215-133

Alarm factor

Demand

Upper
limit
(H)

Example of a display

Alarm factor

Voltageharmonic
nthcontent

Detection display
Alarm setting value

Example of a display

Upper
limit
(H)

Alarm setting
Value

Detection
display

Setting as 11th content of bar graph(It distinguishes in an under-bar)

Current
harmonic
5th
conversion
content

Upper
limit
(H)

Voltage

Upper
limit
(H)

Alarm setting
Value

Setting as 5th conversion content ofbar graph (It distinguishes in anunderbar)

Current
harmonic
nthcontent

Upper
limit
(H)

Voltage

Alarm setting
value

Upper
limit
(H)

Detection
Display

Detection
displayUpper limit (H)alarm set point

Lower limit (L)alarm set point

Detection
display

Setting as 3th content of bar graph(It distinguishes in an underbar)

Current
distortion
factor

Upper
limit
(H)

Voltage

Lower
limit
(L)
Detection
Display

Alarm setting
valueUpper limit (H)alarm set point

Lower limit (L)alarm set point

Detection
display

18

SQLC-215-133

4.2.3 Setting display
It is the display at the case of various setting. There are three types of setting modes according to the contents
of a setting.
Refer to "5. Setting" for the operation in setting mode, and the detailed contents of a setting.
① Setting mode 1
Setting of a measurement display element, an
alarm output, and an alarm value is mainly
performed.
And, an alarm output can be tested in thissetting mode.

New setting
Now setting
Setting No.

Setting mode 1 (No.111 Display pattern)② Setting mode 2
Setting of measurement range, analog output,
pulse output, and measurement display ON/OFF
is mainly performed.
And, initialization of a setting value andreset of watt-hour integrated value can be
performed in this setting mode.

New setting
Now setting
Setting No.
Setting factor
Setting mode 2 (No.211 Voltage range)③ Setting mode 3
Setting of an input circuit and tidal current
measurement is mainly performed.
And, analog output can be adjusted in thissetting mode.

New setting
Now setting
Setting No.

Setting mode 3(No.311 Input circuit phase line change)

19

SQLC-215-133

4.3 Operation
4.3.1 The main monitor display-element change
The measurement display element of the main monitor is changed. A change is performed by ＋ －.
This operation can be performed also except a general measurement display (harmonic measurement display, maximum display
and minimum display). However, in a harmonic measurement display, a sub-monitor changes with a harmonic order. (Themain monitor is distortion-factor fixation.)
After changing a measurement display element, if a switch is not operated for 10 minutes, it will return to the original
measurement display element automatically.
In a harmonic measurement display and maximum display and minimum display, even if a switch is not operated for 10
minutes, it does not return to the original display.
Setting can perform same operation in external operation input. Please refer to "5.3.2 Setting mode 2, (4) external
operation input setting" about the setting method. Please refer to "6.3 Option specification" about external operation
input.
A(L2) display

No operation 10 minute

－

－

＋

＋

＋

＋

－

－

Main monitor A(L1)Wh
V(L1L2) W
display
display
display
Main monitor A(L2)(Default display)

Main monitor A(L3)4.3.2 Phase (line) display change
A phase (line) display of current or voltage is changed. (Everything which is being indicated)
A change is performed by DISPLAY . This operation can be performed also except a general measurement display (harmonicmeasurement display, maximum display and minimum display).
In addition, after changing a phase (line) display, if a switch is not operated for 10 minutes, it will return tothe original phase (line) display automatically.
In a harmonic measurement display and maximum display and minimum display, even if a switch is not operated for 10
minutes, it does not return to the original display.
Setting can perform same operation in external operation input. Please refer to "5.3.2 Setting mode 2, (4) external
operation input setting" about the setting method. Please refer to "6.3 Option specification" about external operation
input.
A(L2) display

No operation 10 minute

DISPLAY

V(L1L2)
display

DISPLAYDISPLAYWh display

W display

A(L2)，V(L1L2) (Default display)

A(L3)，V(L2L3)

20

A(L1)，V(L3L1)

SQLC-215-133

4.3.3 Harmonic measurement display change
A general measurement display and a harmonic measurement display are changed. A change is performed by MODE .
Whenever it pushes a switch, it changes as follows. General measurement display → Current harmonic measurement
display → Voltage harmonic measurement display → General measurement display ･･･This operation can also perform the maximum display and minimum display.
In that case, it changes in order of a next. General measurement maximum and the minimum display → Current harmonicmeasurement maximum display → Voltage harmonic measurement maximum display → General measurement maximum and theminimum display ･･･
In addition, about this operation, even if it does not do switch operation for 10 minutes, it does not return to theoriginal display.
Distortion-factor
displayA(L2) display

Distortion-factor
display

MODE

V(L1L2)
display

Orderdisplay

Wh display
W display

General measurement display.

MODE

MODE

Content
display

Effective-value
display

Orderdisplay

Content
display

Effective-value
display

Current harmonic measurement display. Voltage harmonic measurement display.

4.3.4 Watt-hour enlarged display
In case electric energy is being displayed by the general measurement display, an electric-energy display is usuallychanged to a display (5 digits of integers), and an enlarged display (2 digits integer + below decimal point, 3 digits).
A change is performed by SET .
After an enlarged display, if a switch is not operated for 10 minutes, it returns to the usual display automatically.
<Caution > If it continues pushing SET 3 seconds or more, it will become the setting mode 1. (An electric-energy
display does not change)
A(L2) display

No operation 10 minute

SET

SET

W display Wh display
V(L1L2) display
General display

Watt-hour enlarged display
Enlarged display

21

SQLC-215-133

4.3.5 Setting value check
A voltage range (VT ratio), a current range (CT ratio), and an alarm-output set point are checked.
Check is RESET/SHIFT and － are pushed simultaneously and performed.
The change of a set point is carries out by ＋ and － .
This operation can be performed also except a general measurement display (harmonic measurement display, maximum display
and minimum display). DISPLAY is pushed in case it returns to the original measurement display.
And, if a switch is not operated for 10 seconds after a set point check, it will return to the original measurement
display automatically.
A(L2) display

No operation
10 minute

VT ratioRESET/
＋ －
SHIFT

＋

－

DISPLAY

W display
V(L1L2) display
Wh display

Factor display

General display＋＋
Currentrange(CT ratio)

Input-rating
voltage display

Set point check
Voltage range (VT ratio)－－

Alarm output 2
Time interval，Contact delay
time

＋

－
Alarm output 2
factor，
Return method

＋

22

－
Alarm output 1
Time interval，Contact delay
time＋

Alarm output 1
factor ，Return method－

SQLC-215-133

4.3.6 Setting mode
Various kinds of setting are performed. Setting mode is three types, and operations are different.
DISPLAY is pushed in case it returns to the original measurement display. And, if a switch is not operated for 10
minute after a set point check, it will return to the original measurement display automatically.
Operation and the contents of setting (detail) in setting mode, please refer to "5 Setting".
・Setting mode 1：Press SET for longer than 3 seconds.
・Setting mode 2：Press SET and RESET/SHIFT together for longer than 3 seconds.
・Setting mode 3：Press SET and DISPLAY together for longer than 3 seconds.
<Reference> This operation can be performed also except a general measurement display (harmonic measurement display,
maximum display and minimum display).
Setting No.

New setting

Current setting

SET 3 seconds
SET ＋ RESET/SHIFT 3 seconds

SET ＋ DISPLAY 3 seconds

Measurement display

Setting mode 1
Setting factor
Current setting
New setting

DISPLAY
Return tomeasurement
display
Setting No.
Setting mode 2
Setting No.

Current setting

New setting

Setting mode 3

23

SQLC-215-133

4.3.7 Reset
Various kinds of reset are performed. The kind of reset is as follows and operations are different, respectively.
Reset of watt-hour integrated value (zero clear)，
Reset of maximum value and minimum value (it updates to the instantaneous value at the time)，
Alarm-output reset (OFF of an alarm output (at the case of manual reset setting)).
And, the operation from each measurement display constitutes conditions at each reset.
(1) Watt-hour integrated-value reset
It resets by package about the integrated value of various watt-hour. Watt-hour reset is performed in the setting
mode 2. In detail explanation in the setting mode 2, please refer to "5.3.2 Setting mode 2".
① Press SET and RESET/SHIFT together for longer than 3 seconds. It goes into the setting mode 2.
② MODE is pushed until setting is set to No.271. Further RESET/SHIFT is pushed once and it is made awatt-hour-reset display.
③ Press SET for longer than 3 seconds.
④ DISPLAY is pushed and it returns to a measurement display.
New setting

SET ＋ RESET/SHIFT
3 second

Measurement display

Setting factor

New setting

Present setting

No.211MODE

DISPLAYRESET/SHIFT×ｎ(To setting No.272)

Setting No.

SET 3 second
Wh and varh
display byturns.

Wh and varh
display by turns.

No.272（Clear completion）

It displays by
clear completion.

Setting mode 2

24

Setting No.
No.272（Before clear）

SQLC-215-133(2) Reset of maximum value and minimum value
Reset of the various measurement values of maximum value and minimum value is performed.
This reset has two types of methods. (How to perform according to a measurement factor individual. How to reset
all maximum values and minimum values by package.)a) Individual reset
Reset of only a certain differential maximum value or the minimum value is performed. Other maximum values andminimum values are not reset by this operation.
① A measurement factor to reset is displayed. (General measurement maximum value, minimum value, or Current,
voltage harmonic measurement maximum display)② Press RESET/SHIFT for longer than 1 seconds.
<Caution> An alarm output will be reset if this operation is performed by instant measurement display.
Please be sure to perform this operation after displaying the maximum value and a minimum value
measurement factor to make it reset.
Instant display

The range of minimum to maximum isdisplayed in bar graph.

RESET/SHIFT
1 second

Minimum display

Maximum display

Maximum, Minimum measurement display(General measurement maximum and minimum display)

Minimum display(Reset)Instant display

Maximum display(Reset)Reset completionb) Package reset
Reset of all the maximum values and minimum value is performed.
In addition, setting can perform same operation in external operation input.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method,
Please refer to "6.3 Option" for the external operation input,① Press RESET/SHIFT and ＋ together for longer than 1 seconds. By the maximum and the minimum measurement
display (The general measurement maximum and minimum display or current, voltage harmonic measurement maximum
display).
The range of minimum to maximum isdisplayed in bar graph.
Instant display

RESET/SHIFT ＋ ＋
1 second

Minimum display

Maximum display

Maximum, Minimum measurement display(General measurement maximum and minimum display)

25

Minimum display(Reset)Instant display
Reset completion

Maximum display(Reset)SQLC-215-133(3) Alarm reset
In case an alarm return method is set to “HOLD (manual return)”, an alarm output is reset (output OFF).
(With an alarm-output option)
However, an output is not turned off by this operation, in case an alarm continues and it has caused.
And, this operation is unnecessary in case setting as "AUTO (automatic return)” in alarm return method. (By which
an output is also OFF according to an alarm return.)
In case the number of alarm outputs is two, both outputs are reset (output OFF) by this operation. (Return operationthat comes out individually cannot be performed.)
In addition, setting can perform same operation in external operation input.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method,
Please refer to "6.3 Option" for the external operation input,① By instant measurement display (a general measurement display or current, voltage harmonic measurement display),
RESET/SHIFT is pushed 1 second or more.
<Caution> If this operation is performed by the maximum and the minimum measurement display, the maximum value andminimum value of the measurement factor currently displayed will be reset.
Please be sure to perform this operation in the state of an instant measurement display.
Alarm detection (Hold)RESET/SHIFT
1 second

Instant measurement display(General measurement display)

Reset completion

26

SQLC-215-133

5. Setting
5.1 Function table
This product has each function setting with a front switch.
Setting mode 1. Function table (1)SetFunction
No.

Functional description

Current
input

Default value

ImportantPage
setting111

Display pattern

Set the display combination pattern of thedigital 4 displays and bar graph display.

○

Pattern 1

○

41-43

112

Main monitor

Set the display factor of digital main monitor.

○

A(L2)

○

41-43

113

sub-monitor (left)

Set the display factor of digital sub-monitor(left).

○

V(L1L2)

○

41-43

114

sub-monitor (center)Set the display factor of digital sub-monitor(center).

○

W

○

41-43

115

sub-monitor (right)

Set the display factor of digital sub-monitor(right).

○

Wh

○

41-43

116

Bar graph

Set the display factor of bar graph.

○

A(L2)

○

41-43

121AL
(12)

Alarm 1 factor

Set the output factor of alarm 1.

○

DA

○

44

122AL
(12)

Alarm 1 return method

Set the output action at the caseof reset of alarm 1.

○

Automatic reset

44

123AL
(12)

Alarm 1 mask time(Contact delay time)

Set the contact delay time of alarm 1.

○

0 second

44

124AL
(12)

Alarm 1 test

Output test of alarm 1 is performed.

○

－

44

125AL
(12)

Alarm 2 factor

Set the output factor of alarm 2.

○

DA

126AL
(12)

Alarm 2 return method

Set the output action at the caseof reset of alarm 2.

○

Automatic reset

44

127AL
(12)

Alarm 2 mask time(Contact delay time)

Set the contact delay time of alarm 2.

○

0 second

44

128AL
(12)

Alarm 2 test

Output test of alarm 2 is performed.

○

－

44

131H

Demand current upper
limit

Set the high-alarm value of demand current.

○

80%(Full scale=100%)

132

Demand current timeinterval

Set time interval of demand current.

○

0 second

133H

Demand active power
upper limit

Set the high-alarm value of demand active
power.

OFF (No operation)

134

Demand active power
time interval

Set the time interval of demand active power.

0 second

135

Demand active power
operating method

Set the operating method of demand active
power.

136

Power-factor
operating method

Set the operating method of power-factor
measurement.

Operating system
according withbimetallic type.
Instant
measurement

Note(12) A setting item is not displayed in case there is no corresponding option.

27

○

44

45,46
○

45,46
45,46

○

45,46

○

45,46
45,46

SQLC-215-133

Setting mode 1. Function table (2)SetCurrent
Important
Function
Functional description
Default value
Page
No.
input
setting
Current
Set the high-alarm value of current distortion
141H distortion-factor○
OFF (No operation)
46,47
factor.
upper limit
Current 5th
Set the high-alarm value of current 5th
142H conversion content
46,47○
OFF (No operation)
conversion content.
upper limit
Current n-th
143
Set the order of n-th current content.
○
5th
46,47
content factor
144H
145H

146H
147
148H
149
14A

Current n-th content
upper limit
Voltagedistortion-factor
upper limit
Voltage 5th
conversion content
upper limit
Voltage n-thcontent factor

Set the high-alarm value of current n-thcontent.

Voltage n-th content
upper limit
5th conversion
detection
characteristics
Average value timeinterval○

OFF (No operation)

46,47

Set this high-alarm value of voltage
distortion factor.

OFF (No operation)

46,47

Set the high-alarm value of voltage 5th
conversion content.

OFF (No operation)

46,47

Set the order of n-th voltage content.

5th

46,47

Set the high-alarm value of voltage n-thcontent.

OFF (No operation)

46,47

Set the detection characteristic of 5thconversion content.

○

Inverse-timedelay mode

46,47

Set the average value detection time intervalof harmonic.

○

0 minute

46,47

151H

Instant measurement
voltage upper limit

Set the high-alarm value of instant voltage.

OFF (No operation)

47

152L

Instant measurement
voltage lower limit

Set the low-alarm value of instant voltage.

OFF (No operation)

47

171

Backlight action

Set the ON/OFF of backlight.

○

AUTO OFF48

172

Backlight brightness

Set the brightness of backlight.

○3（Middle）

48

28

SQLC-215-133

Setting mode 2. Function table (1)SetFunction
No.

Functional description

Current
input

Default value

ImportantPage
setting

6600V (14)

○

49-51

○

100.0A

○

49-51

○

100.0A

49-51

Active power polarity Set the swing display of active power meter.

One-side swing

49-51

215

Active power range

Set the full scale of active power meter.

1200kW (15)

49-51

216

Reactive power range

Set the full scale of reactive power meter.

600kvar (15)

49-51

217

Power-factor range

Set the full scale of power-factor meter. And,
set this output range of analog output.

0.5‐1‐0.5

49-51

218

Frequency range

Set the full scale of frequency meter.
And, set this output range of analog output.

45‐65Hz

49-51

221A
(13)

AO1 output factor

Set the output factor of AO1 (analog output 1).

○

A(L2)

○

52,53

222A
(13)

AO2 output factor

Set the output factor of AO2 (analog output 2).

○

V(L1L2)

○

52,53

223A
(13)

AO3 output factor

Set the output factor of AO3 (analog output 3).

○

W

○

52,53

224A
(13)

AO4 output factor

Set the output factor of AO4 (analog output 4).

○

cosφ

○

52,53

225A
(13)

Current output
peculiar sensitivity

Set the analog output sensitivity(％ of output for input) of current.

○

100.0％

52,53

226A
(13)

Active power-output
peculiar sensitivity
Reactive power
output peculiar
sensitivity

Set the analog output sensitivity(％ of output for input) of active power.

100.0％

52,53

Set the analog output sensitivity(％ of output for input) of reactive power.

50.0％

52,53

OFF (No operation)

52,53

211

Voltage range

Set the voltage-measurement range (VT ratio).

212

Current range

Set the current-measurement range (CT ratio).

213

Current display
peculiar sensitivity

Set the full scale of current meter.

214

227A
(13)
228A
(13)

Low input cut

Set the function which makes a lower limit theoutput at the case of a minute input (adequate
to 0.5％ or less) in analog output.

241P
(13)

PO1 factor

Set the output factor of PO1 (pulse output 1).

Wh

242P
PO1 pulse unit
(13)(16)

Set the output pulse unit of PO1(pulse output 1).

10kWh/p

243P(13)

Set the output factor of PO2 (pulse output 2).

Wh

Set the output pulse unit of PO2(pulse output 2).

10kWh/p

PO2 factor

244P
PO2 pulse unit
(13)(16)

○

○

53
53

○

53
53

251
(13)

External operation
input 1 function

Set the function of the external operation
input 1.

○

Alarm reset

○

54

252
(13)

External operation
input 2 function

Set the function of the external operation
input 2.

○

Maximum /Minimum reset○

54

Note(13) A setting item is not displayed in case there is no corresponding option.
Note(14) It is set to “220V” in 220V input.
Note(15) It is set to "40.0kW" and "20.00kvar" in 220V input.
Note(16) It is set to “0.1kWh/p” in 220V input.

29

SQLC-215-133

Setting mode 2. Function table (2)SetFunction
No.

Functional description

261

Voltage ON/OFF

Set the ON/OFF of voltage measurement display.

262

Current ON/OFF

Set the ON/OFF of current measurement display.

263

Active power ON/OFF

264

Current
input

Default value

ImportantPage
setting

ON

55

ON

55

Set the ON/OFF of active power measurement
display.

ON

55

Reactive power
ON/OFF

Set the ON/OFF of reactive power measurement
display.

ON

55

265

Power-factor ON/OFF

Set the ON/OFF of power-factor measurement
display.

ON

55

266

Frequency ON/OFF

Set the ON/OFF of frequency measurement
display.

ON

55

267

Power-receiving
watt-hour ON/OFF

Set the ON/OFF of power-receiving watt-hourmeasurement display.

ON

55

Power transmission
var-hour ON/OFF
Power-receiving
var-hour (LAG,LEAD)
ON/OFF
Power transmission
var-hour (LAG,LEAD)
ON/OFF
Harmonic current
ON/OFF

Set the ON/OFF of power transmission var-hourmeasurement display.

ON

55

Set the ON/OFF of power-receiving var-hour(LAG,LEAD) measurement display.

ON

55

Set the ON/OFF of power transmission var-hour(LAG,LEAD) measurement display.

ON

55

ON

55

26C

Harmonic voltage
ON/OFF

Set the ON/OFF of harmonic voltage measurement
display.

ON

55

271

set value
initialization

All set values are initialized.
(Return to initial set value)

－

55

272

Watt-hour reset

Integrated value is cleared by package about
each electric energy.

－

55

268
269

26A
26B

Set the ON/OFF of harmonic current measurement
display.

30

○

○

○

SQLC-215-133

Setting mode 3. Function table
Set
Function
No.

Functional description

Current
input

Default value○

3φ3W (Fixation)

○

56

110V

○

56

311

Input circuit phaseline change

2VT･3CT product is 3-phase 3 line (3φ3W)
fixation.

312

Input voltage

Set the input voltage.

321

Measurement deadband

Set the dead band of measurement display.

322

Tidal current
measurement

Set the general measurement or tidal current
measurement which was conscious of powertransmission / power receiving, in measurementof reactive power and power-factor.

Analog output
331(17)(18) specification○

ImportantPage
setting

0.0％

57

Generalmeasurement

57

Set the analog output specification.

○

DC1‐5V or DC0‐5V(Designation )

58

341
(17)

AO1 BIAS adjustment

Set the BIAS value of AO1 (Analog output 1).

○

0.0％

58

342
(17)

AO1 SPAN adjustment

Set the SPAN value of AO1 (Analog output 1).

○

100.0％

58

343
(17)

AO2 BIAS adjustment

Set the BIAS value of AO2 (Analog output 2).

○

0.0％

58

344
(17)

AO2 SPAN adjustment

Set the SPAN value of AO2 (Analog output 2).

○

100.0％

58

345
(17)

AO3 BIAS adjustment

Set the BIAS value of AO3 (Analog output 3).

○

0.0％

58

346
(17)

AO3 SPAN adjustment

Set the SPAN value of AO3 (Analog output 3).

○

100.0％

58

347
(17)

AO4 BIAS adjustment

Set the BIAS value of AO4 (Analog output 4).

○

0.0％

58

348
(17)

AO4 SPAN adjustment

Set the SPAN value of AO4 (Analog output 4).

○

100.0％

58

Note(17) A setting item is not displayed in case there is no corresponding option.
Note(18) A setting item is not displayed other than analog output DC0‐5V or DC1‐5V specification.

31

SQLC-215-133

5.2 Setting table
A setting item changes by the specification of a product, or the existence of an option.
(1) Important setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the measurement
range of voltmeter(211)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Select a measurement range by ＋ and －Press SET
Selected measurement range is entered

Set the measurement
range of ammeter(212)

Set the display
combination(111)

Press DISPLAY

Page

49-51

Returns to display mode.

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Select a measuring range by ＋ and －Press SET
(212)
Selected measuring range is entered
Press DISPLAY
Returns to display mode.
Select the display combination by ＋ and －Press SET for longer than 3 seconds(111)
Press SET
Selected display combination is entered
Press DISPLAY

49-51

41-43

Returns to display mode.
Set the output
factor of analog
output 1 (AO1).
(221A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Select an output factor by ＋ and －Press SET
Selected output factor is entered

Press DISPLAY

52,53

Returns to display mode.

Set the output
factor of analog
output 2 (AO2).
(222A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Select an output factor by ＋ and －Press SET(222A)
Selected output factor is entered
Press DISPLAY
Returns to display mode.

52,53

Set the output
factor of analog
output 3 (AO3).
(223A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Select an output factor by ＋ and －(222A)(223A)
Press SET
Selected output factor is entered
Press DISPLAY
Returns todisplay mode.

52,53

Set the output
factor of analog
output 4 (AO4).
(224A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)
Select an output factor by ＋ and －Press SET
Selected output factor is entered

Set the output
factor of pulse
output 1 (PO1).
(241P)

Set the output
factor of pulse
output 2 (PO2).
(243P)

Press DISPLAY

Returns to display mode.

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Select an output factor by ＋ and －Press SET
(241P)
Selected output factor is entered
Press DISPLAY
Returns to display mode.
Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press RESET/SHIFT
Press RESET/SHIFT(241P)(242P)(243P)
Select an output factor by ＋ and －Press SET
Selected output factor is entered

Press DISPLAY

32

52,53

Returns to display mode.

53

53

SQLC-215-133

Items
Set the factor ofalarm output 1.
(121AL)

Setting and operation procedures
Press MODE
Press SET for longer than 3 seconds(111)(121AL)
Select an output factor by ＋ and －Press SET
Selected factor is entered
Press DISPLAY

Set the factor ofalarm output 2.
(125AL)

44

Returns to display mode.

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(123AL)(124AL)(125AL)
Press SET
Selected factor is entered
Select an factor by ＋ and －Press DISPLAY

Page

44

Returns to display mode.

Set the external
operation input 1
function.
(251)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press MODE
Select an function by ＋ and －Press SET
(241P)(251)
Selected function is entered
Press DISPLAY
Returns to display mode.

54

Set the external
operation input 2function.
(252)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press MODE
Press RESET/SHIFT
Select an function by ＋ and －(241P)(251)(252)
Press SET
Selected function is entered
Press DISPLAY
Returns to display mode.

54

Set the time intervalof demand current.
(132)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Select an time interval by ＋ and －Press SET
(132)
Selected time interval is entered
Press DISPLAY
Returns to display mode.

Set the time intervalof demand active
power.
(134)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)
Select an time interval by ＋ and －Press SET
Selected time interval is entered
Press DISPLAY

45,46

45,46

Returns to display mode.

Set the operation
method of demand
active power.
(135)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)
Press RESET/SHIFT
Select an operation method by ＋ and －Press SET
(135)
Selected operation method is entered
Press DISPLAY
Returns to display mode.

Set the phase wire of
input circuit.
(311)

2VT･3CT product is 3-phase 3 line (3φ3W) fixation.

56

Press RESET/SHIFT
Press SET and DISPLAY together for longer than 3 seconds
(311)
(312)
Select an input voltage by ＋ and －Press SET56

Set the input
voltage.
(312)

Selected input voltage is entered

Press DISPLAY

33

Returns to display mode.

45,46

SQLC-215-133(2) A combination except a display pattern.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the display
factor of main
monitor.
(112)

Set the display
factor of
sub-monitor (left).
(113)

Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Select an display factor by ＋ and －Press SET
Selected display factor is entered

Press DISPLAY

41-43

Returns to display mode.

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Select an display factor by ＋ and －Press SET
Selected display factor is entered

Page

Press DISPLAY

41-43

Returns to display mode.

Set the display
factor of
sub-monitor(center).
(114)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Press RESET/SHIFT
Select an display factor by ＋ and －Press SET
(114)
Press DISPLAY
Returns to display mode.
Selected display factor is entered

41-43

Set the display
factor of
sub-monitor (right).
(115)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Press RESET/SHIFT
Press RESET/SHIFT
Select an display factor by ＋ and －
(114)
(115)
Press SET
Selected display factor is entered
Press DISPLAY
Returns todisplay mode.

41-43

Set the display
factor of bar graph.
(116)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(114)
(115)
(116)
Select an display factor by ＋ and －(If a sub-monitor is selected, an underbar will
be displayed on the bottom of a digital display.)
Press SET

Selected display factor is entered

Press DISPLAY

Returns todisplay mode.

(3) Setting of active power polarity and measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Setting of
measurement rangeof active power.
(215)

41-43

Page

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Select an one-side(P)/both-side(-) deflection by ＋ and －Press SET
49-51
Selected deflection is entered

Press RESET/SHIFT(215)
Select a measuring range by ＋ and －Press SET
Selected measuring range is entered

34

Press DISPLAY

Returns to display mode.

SQLC-215-133(4) Setting of reactive power measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Setting of
measurement range ofreactive power.
(216)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
(215)
Press RESET/SHIFT
Select a measuring range by ＋ and －Press SET(216)
Selected measuring range is entered
Press DISPLAY
Returns to display mode.

(5) Setting of Wh (varh) output pulse unit.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the pulse unitof pulse output 1(Po1).
(242P)

Set the pulse unitof pulse output 2(Po2).
(244P)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Select a output pulse unit by ＋ and －Press MODE
(241P)(242P)
Press SET
Selected output pulse unit is entered
Press DISPLAY
Returns todisplay mode.
Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(241P)(242P)(243P)(244P)
Select a output pulse unit by ＋ and －Press SET
Selected output pulse unit is entered

Press DISPLAY

Set the measurement
range of frequency.
(218)

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(213)
(214)
(215)
(216)
Select a measuring range by ＋ and －Press SET
Press RESET/SHIFT(217)
Selected measuring range is entered
Press DISPLAY
Returns to display mode.
Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
(215)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(216)
(217)
(218)
Select a measuring range by ＋ and －Press SET
Selected measuring range is entered

Press DISPLAY

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Select a display sensitivity by ＋ and －
(212)
(213)
Press SET
Selected display sensitivity is entered
Press DISPLAY
Returns to display mode.

35

Page

53

53

Page

49-51

49-51

Returns to display mode.

(7) Setting of current display intrinsic sensitivity.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the display
sensitivity (％ ofa display to an
input) of current.
(213)

49-51

Returns to display mode.

(6) Setting of active power-factor and frequency measurement range
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Set the measurement
range of active
power-factor.
(217)

Page

Page

49-51

SQLC-215-133(8) Setting of analog output.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the analog
output sensitivity(％ of an output toan input) of current.
(225A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Select a output sensitivity by ＋ and －Press SET
Selected output sensitivity is entered

Press DISPLAY

Page

52,53

Returns to display mode.

Set the analog
output sensitivity(％ of an output toan input) of active
power.
(226A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Press RESET/SHIFT
Select a output sensitivity by ＋ and －Press SET
(226A)
Selected output sensitivity is entered
Press DISPLAY
Returns to display mode.

Set the analog
output sensitivity(％ of an output toan input) of reactive
power.
(227A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
52,53
Press RESET/SHIFT
Select a output sensitivity by ＋ and －Press RESET/SHIFT(226A)(227A)
Press SET
Selected output sensitivity is entered
Press DISPLAY
Returns to display
mode.

Set the output cut
function at thecase in minute input(0.5％ or less).
(At the case ofanalog output)(228A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(226A)(227A)(228A)
Select a low Input cut ON/OFF by ＋ and －Press SET
Selected action is entered

Press DISPLAY

36

Returns to display mode.

52,53

52,53

SQLC-215-133(9) Setting of alarm output.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the return
method of alarm
output 1.
(122AL)Set the contact
delay time of alarm
output 1.
(123AL)Set the return
method of alarm
output 2.
(126AL)

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Select a return method by ＋ and －Press SETThe selected return method is entered

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(123AL)(124AL)(125AL)(126AL)
Press SET
Select a return method by ＋ and －Press DISPLAY

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(123AL)(124AL)(125AL)(126AL)
Press RESET/SHIFT
Select an contact delay time by ＋ and －Press SET
(127AL)The selected contact delay time is entered
Press DISPLAY
Returns to display mode.

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

44

44

Returns to display mode.

(10) Demand measurement (current, active power) setting.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the high-alarm
value of demandcurrent.
(131H)

44

Returns to display mode.

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Select an contact delay time by ＋ and －Press SET
(123AL)The selected contact delay time is entered
Press DISPLAY
Returns to display mode.

The selected return method is entered

Set the contact
delay time of alarm
output 2.
(127AL)

Press DISPLAY

Page

44

Page

45,46

Returns to display mode.

Set the high-alarm
value of demand
active power.
(133H)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Select a high-alarm value by ＋ and －(132)(133H)
Press SET
Selected high-alarm value is entered
Press DISPLAY
Returns todisplay mode.

45,46

Set the operation
method ofpower-factor
measurement.
(136)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)
Press RESET/SHIFT
Press RESET/SHIFT
Select a operation method by ＋ and －
(135)
(136)
Press SET
Selected operation method is entered
Press DISPLAY
Returns todisplay mode.

45,46

37

SQLC-215-133(11) Harmonic measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the high-alarm
value of current
distortion factor.
(141H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

Page

46,47

Returns to display mode.

Set the high-alarm
value of 5th current
conversion content.
(142H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Select a high-alarm value by ＋ and －Press SET
(142H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

46,47

Set the order ofn-th current
content.
(143)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Select a order by ＋ and －Press SET
(142H)(143)
Selected order is entered
Press DISPLAY
Returns to display mode.

46,47

Set the high-alarm
value of n-th
current content.
(144H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Set the high-alarm
value of voltage
distortion factor.
(145H)

Press DISPLAY

Returns to display mode.

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

46,47

46,47

Returns to display mode.

Set the high-alarm
value of voltage 5th
conversion content.
(146H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Select a high-alarm value by ＋ and －Press SET
(146H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

46,47

Set the order ofn-th voltage
content.
(147)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Select a order by ＋ and －Press SET
(146H)(147)
Selected order is entered
Press DISPLAY
Returns to display mode.

46,47

Set the high-alarm
value of n-th
voltage content.
(148H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(146H)(147)(148H)
Select a factor by ＋ and －Press SET
Selected factor is entered
Press DISPLAY

Returns to display mode.

38

46,47

SQLC-215-133

Items

Setting and operation procedures

Set the detectedcharacteristics of
5th conversion
content.
(149)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(146H)(147)(148H)(149)
Press SET
Select a detected characteristics by ＋ and －Selected detected characteristics is entered

Set the averagetime interval.
(14A)

Press DISPLAY

Page

46,47

Returns to display mode.

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(146H)(147)(148H)(149)
Press RESET/SHIFT
Select a time interval by ＋ and －Press SET
(14A)
Selected time interval is entered
Press DISPLAY
Returns to display mode.

(12) Instant measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

46,47

Page

Set the high-alarm
value of instantvoltage.
(151H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Select a high-alarm value by ＋ and －Press SET
Press MODE
(151H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

47

Set the low-alarm
value of instantvoltage.
(152L)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press RESET/SHIFT
Select a low-alarm value by ＋ and －(151H)(152L)
Press SET
Selected low-alarm value is entered
Press DISPLAY
Returns todisplay mode.

47(13) Backlight setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the action of
backlight.
(171)

Set the brightnessof backlight.
(172)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Select a backlight action by ＋ and －Press SET
(151H)(171)
Selected backlight action is entered
Press DISPLAY
Returns to display mode.
Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Press RESET/SHIFT
Select a brightness by ＋ and －(151H)(171)(172)
Press SET
Selected backlight brightness is entered
Press DISPLAY
Returns to display mode.

39

Page

48

48

SQLC-215-133(14) Other, measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the dead bandof measurement
display.
(321)
Set the tidal
current measurementof reactive power
and power-factor.
(322)

Press MODE
Press SET and DISPLAY together for longer than 3 seconds
(311)
(321)
Select of dead band value in measurement display by ＋ and －Press SETThe value of dead band in display is entered

Press DISPLAY

5.3 Setting in detail explanation
5.3.1 Setting mode 1SET
3 seconds
111 to 116
Display setting
MODE
DISPLAY

121AL to 128AL
Alarm output setting
MODE

DISPLAY

Display mode
(Measurement
display)

131H to 136
Demand detection setting
MODE

DISPLAY

141H to 14A
Harmonic detection setting
MODE

DISPLAY

151H to 152L
Instant measurement detection setting
MODE

DISPLAY

171 to 172
Backlight setting
MODE

Setting mode 1

Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, an alarm output is not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

40

57

Returns to display mode.

Press MODE
Press SET and DISPLAY together for longer than 3 seconds
(311)
(321)
Press RESET/SHIFT
Select a tidal current measurement ON/OFF by ＋ and －(322)
Press SET
Selected action is entered
Press DISPLAY
Returns to display mode.

DISPLAY

Page

57

SQLC-215-133(1) 111 to 116 Display combination setting● Voltage, current input (19)№

Pattern №

Main monitor

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18

Pattern 1
Pattern 2
Pattern 3
Pattern 4
Pattern 5
Pattern 6
Pattern 7
Pattern 8
Pattern 9
Pattern 10
Pattern 11
Pattern 12
Pattern 13
Pattern 14
Pattern 15
Pattern 16
Pattern 17
Pattern 18

A(L2)
A(L2)
A(L2)
DA(L2)
DA(L2)
DA(L2)
W
W
W
DW
DW
A(L2)
A(L2)
W
A(L2)
V(L1L2)
A(L1)
A(L1)

Sub-monitor(Left)
V(L1L2)
V(L1L2)
V(L1L2)A(L2)A(L2)
V(L1L2)
V(L1L2)
V(L1L2)
V(L1L2)
V(L1L2)
V(L1L2)
cosφ
var
cosφ
A(L1)
V(L2L3)－－Sub-monitor(Center)
W
W
W
V(L1L2)
V(L1L2)
W
A(L2)
A(L2)
A(L2)
W
A(L2)
W
W
var
A(L3)
V(L3L1)
V(L1L2)
V(L1L2)Sub-monitor(Right)
Wh
cosφ
Hz
W
Wh
cosφ
Wh
cosφ
Hz
Wh
cosφ
Wh
Wh
Wh
Wh
Hz
W
Wh

Bar graph
A(L2)
A(L2)
A(L2)
DA(L2)
DA(L2)
DA(L2)
W
W
W
DW
DW
A(L2)
A(L2)
W
A(L2)
V(L1L2)
A(L1)
A(L1)

Note(19) Current input is pattern 15 only. (However, sub-monitor (right) is displayed blank.)● Displays set factor
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，var，cosφ，Hz，Main monitor
Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，Distortion factor(A,V)
Sub-monitor
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，W，var，cosφ(Left)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，var，Sub-monitor
varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，Harmonic 5th conversion content(A,V)，(Center)
Harmonic nth content(A,V)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，cosφ，Hz，Sub-monitor
Wh，-Wh，Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，(Right)Harmonic nth effective value(A,V)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，var，cosφ，Hz，Distortion factor(A,V)，Harmonic 5th conversion content(A,V)，Harmonic nth content(A,V)，
Bar graph
Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，
Harmonic nth effective value(A,V)● Phase (line) change (3-phase 3-wire) ● Measurement factor change (3-phase 3-wire, Measurement display mode)(20)
V(L1L2) V(L2L3) V(L3L1)
A(L1)
A(L2)
A(L3)
DA(L1)
V(L1L2) V(L2L3) V(L3L1)
A(L2)

A(L3)

A(L1)

Note(20) Press DISPLAY , Voltage
and current are replacedat the same time.

DA(L2)

DA(L3)

Wh-Wh-varh(LEAD)

W
varh(LAG)

var

cosφ

varh(LEAD)

Hz-varh(LAG)

Nothing● Measurement factor change (Harmonic measurement display mode）
Main monitor：Distortion factor (Fixation)
Sub-monitor (Left) ：Harmonic order ｎ
5th conversion
1
Sub-monitor (Center)：Harmonic nth content (Fixation)
Sub-monitor (Right) ：Harmonic nth effective value (Fixation)

41

DW

3

4

5

7

9

11

13

15

SQLC-215-133

Display combination setting
RESET/SHIFT
111
Combination
display

RESET/SH
IFT

112
Main monitor

Pattern 1

A(L2)：5＋＋－

RESET/SH
IFT

113
Sub-monitor(Left)RESET/SH
IFT

V(L1L2)：1

－

＋

114
Sub-monitor(Center)

RESET/SH
IFT

＋

－

＋

A(L3)：6

V(L2L3)：2

DW：11-Wh：15

＋

＋

＋

＋

＋

DA(L1)：7

Pattern 3

＋
＋

－

－

＋

＋

－

＋

－

＋

－
＋

＋

A(L1)：4－

Pattern 17

＋

＋

＋

DA(L3)：9

A(L2)：5＋＋－

Pattern 18
＋

－

－

W：10

－

Manual setting
value：0

＋

＋

－

＋

－

＋

＋

－

＋

－

＋

Nothing：0＋＋

cosφ：9

－

＋

＋

－

－

＋

－

－

－

－

A(L2)：5－

V(L1L2)：1＋－

Nothing：0

－

Wh：16
＋

－

＋
－

＋

－

＋

－

DA(L1)：7＋＋－

V(L3L1)：3

DA(L2)：8

＋

＋

－

－

-Wh：17

A(L1)：4

DA(L3)：9

＋

＋

＋

－

varh(LAG)：18＋＋－

varh(LEAD)：19＋－-varh(LAG)：20
＋

－

－

－

－

V(L3L1)：3
＋

－

DW：11
＋

－

DA(L1)：7

cosφ：12

＋

＋

－

－

Hz：13

－

Default setting－

V(L2L3)：2＋＋

DA(L3)：9V(L1L2)：1
＋

－

＋

－

At the time of
3-phase 3-wire

W：10

A(L3)：6＋

Nothing：0＋－

DA(L2)：8-varh(LEAD)：21＋－

A(L2)：5－

A(L3)：6

V(L2L3)：2－

Sub-monitor：RightA(L1)：4

－

var：8

Hz：14

＋

－

＋

－

A(L1)：4

42－

V(L3L1)：3-varh(LEAD)：16

var：12＋＋-varh(LAG)：15

－

＋

－

V(L2L3)：2

－

＋

－

＋

－

varh(LEAD)：14

W：7

cosφ：13

＋

＋

－

＋

DW：11

At the cases other thana display combination
pattern, it becomesmanual setting.

－A(L3)：6

V(L1L2)：1

varh(LAG)：13－

Sub-monitor：Center－
DA(L2)：8－

Sub-monitor：Left

Nothing：0

var：12V(L3L1)：3－－

Main monitor－

Pattern 2－

116
Bar graphfactorRESET/SH
IFT

Wh：14

W：10

－

115
Sub-monitor(Right)－SQLC-215-133◆ 111 Combination display
Select the factors to be measured and monitored by 4 digital displays
out of combination patterns.
Set values are updated by SET .
◆ 112 to 115 Main monitor, Sub-monitor (left), Sub-monitor (center),
Sub-monitor (right)
Set these items for a display configuration other than combined
patterns. Set values are updated by SET .
◆ 116 Bar graph factor
A factor being monitored in the main monitor is basically displayed
by a bar graph. Set this item for displaying a factor being monitored
on a sub-monitor by bar graph. An underbar is attached to the digital
display of the setting sub-monitor.
Set values are updated by SET .

Setting №
Underbar

43

Setting №

New setting
Now setting

SQLC-215-133(2) 121AL to 128AL Alarm output setting 【With an alarm output option】
Various setting and an output test are performed about an alarm output.
In case contact outputs 1 and 2 are alarm-output specifications, the corresponding alarm output is setting.
121AL
Alarm 1 factor

RESET/SHIFTRESET/SHIFT
128AL
Alarm 2 test

RESET/SHIFT

122AL
Alarm 1
return method

RESET/SHIFT

127AL
Alarm 2 contact
delay timeRESET/SHIFT

123AL
Alarm 1 contact
delay timeRESET/SHIFT

126AL
Alarm 2
return method

RESET/SHIFT

124AL
Alarm 1 test
RESET/SHIFT
125AL
Alarm 2 factor◆ 121AL Alarm 1 factor setting，125AL Alarm 2 factor setting.
Set the output factor of alarms 1 and 2. Selection by ＋ and － , set value is updated by SET .
Initial value ：1 (DA：Demand current)
OFF：Nothing＋＋

1：Demandcurrent－－

2：Demand
power

＋
－

4：Distortion
factor(Current)＋－

5：Harmonic 5th
conversion
content(Current)＋ －－ ＋
10：Voltage

9：Harmonic nth
content(Voltage)

－
＋

－
＋

8：Harmonic 5th
conversion
content(Voltage)－＋

7：Distortion
factor(Voltage)－＋

6：Harmonic nth
content(Current)◆ 122AL Alarm 1 reset method setting,
126AL Alarm 2 reset method setting
Output action at the case of a reset of alarms 1 and 2 can be selectedfrom AUTO (automatic reset) and HOLD (manual reset).
In "AUTO (automatic reset)", an alarm output also constitutes OFF
according to a reset of an alarm. In "HOLD (manual reset)", even afteran alarm reset, an output holds ON. The reset in this case (output OFF)is performed in RESET/SHIFT .
Selection by ＋ and － , set value is updated by SET .
Default setting：AUTO (automatic reset)＋

AUTO(Auto reset)

＋
－

HOLD
(Manual reset)

New setting

Setting №

Currenttti－◆ 123AL Alarm 1 contact delay time, 127AL Alarm 2 contact delay time
Set the contact delay time of alarms 1 and 2. The setting range is 0 to 300 seconds (1-second step).
Selection by ＋ and － , set value is updated by SET .
Default setting：0 second (With no contact delay)◆ 124AL Alarm 1 test, 128AL Alarm 2 test
The output of alarms 1 and 2 is
Setting No.
tested. While pushing SET , anoutput is ON, and if it detaches,
an output switches OFF. In
addition, about a reset of an alarm,
it becomes action united with reset
method setting (automatic reset /manual reset) of each alarm output.

Status of output

Setting No.

SET(Output OFF)

(Output ON)
Alarm 1 test

44

Status of output

SQLC-215-133(3) 131H to 136 Demand detection setting
The following operation method is setting.
Demand current, action of demand, high-alarm value, time-interval, power-factor, demand active power.
131H
Demand current
upper limit

RESET/SHIFT

136
Power-factor
operation method

RESET/
SHIFT

132
Demand current
time interval

RESET/
SHIFT

135
Demand power
operation method

RESET/SHIFT

133H
Demand power
upper limit

RESET/
SHIFT

134
Demand power
time interval

RESET/SHIFT◆ 131H Demand current upper limit. 133H Demand active power upper limit.
Set the high-alarm value of demand current (DA) and demand active power (DW).
The setting range is 5 to 100％ (1％ step). Selection by ＋ and － , set value is updated by SET .
Default setting：80％ (Demand current),
OFF [Non use] (Demand active power)
New setting
Current◆ 132 Demand current time interval. 134 Demand active power time interval.
Set the time interval (95％ time interval) of demand current(DA) anddemand active power(DW). Selection by ＋ and － , set value is updated
by SET .
Default setting：0 second (demand current, demand active power)
0sec(Nothing)

＋
－
－
＋

5sec

4min

5min
－
＋

30min

＋
－
－
＋
＋
－
－
＋

10sec

3min

6min

25min

＋
－
－
＋
＋
－
－
＋

20sec

2min

7min

20min

＋
－

30sec

－

1min

＋
＋

8min

－
－
＋

15min

＋
－
－
＋
＋
－
－
＋

40sec

＋
－

50sec

9min

＋
－

Setting factor

Setting No.

Demand current time interval

10min◆ 135 Demand active power operation method
The operating system of demand (DW) can be selected fromd：Operating systemd (operating system according with bimetallic type：according withdemand) and A (Averaging arithmetic in demand time
bimetallic typedelay：average).
Selection by ＋ and － , set value is updated by SET .
Default setting：d (Operating system according with bimetallic type)＋＋－

A：Averaging
arithmetic indemand time
delay

－

● Demand time delay characteristic (Demand current, Demand active power)
100
95
80

Arithmetic method accordingwith bimetallic type.
(Indication time to 95％
of a final constant value)

Time delay 10 minutes

Demandvalue 60（％）

Averaging arithmetic indemand time delay.
(Averaging time in demand
time delay.)

40

20
0
0

10

20

30

Time（min）

Arithmetic method
Demand current measurement：Arithmetic method according with bimetallic type.
Demand active power measurement：Arithmetic method according with bimetallic type (Initial value).
Or the averaging arithmetic in a demand time interval.
One is selected by setting.
100％ indication time is about 3 times the time delay at the case of the arithmetic method according with bimetallic
type.(In case of 10 minutes/95％ of time-interval, time to reach to 100％ is about 30 minutes.)
Demand measurement is measured to the 2 times of the rated current, and the 2 time of a rated active power.

45

SQLC-215-133◆ 136 Power-factor operation method
The operation method of power-factor measurement can be selected from 0(instant measurement) and 1 (averaging operator in a demand timeinterval).
In case it is set as "1 (averaging operator in a demand time interval)",power-factor measurement is calculated from the operation method of apower demand time interval and a demand active power meter.
Selection by ＋ and － , set value is updated by SET .
Default setting：0 (Instant measurement )＋0：Instantmeasurement＋

1：Averaging operator－

in a demand timeinterval

New setting－

Setting No.

Current setting

Power-factor operation method(4) 141H to 14A Harmonic detection setting
Set the high-alarm value of each harmonic (current, voltage) measurement factor, factor, 5th conversion detection
characteristics, and average time interval is performed.
141H
Current
distortion-factor
upper limit

RESET/SHIFTRESET/SHIFT

142H
Current 5th
conversion content
upper limit
147
Voltage nthcontent factor

RESET/SHIFT

148H
Voltage nthcontent upper
limit

143
Current nth
content factor

RESET/SHIFT

144H
Current nth
content upper
limit

RESET/SHIFT

146H
RESET/
SHIFT

Voltage 5th
conversion
content upper

RESET/SHIFT

5th conversion
detection
characteristics

145H
RESET/
SHIFT

Voltagedistortion-fact
or upper limit

RESET/SHIFT

Average timeinterval149◆ 141H Current distortion-factor upper limit,
145H Voltage distortion-factor upper limit

RESET/SHIFT

14A

Setting factor

New setting

Set the high-alarm value of distortion factor (current, voltage).
Setting range. Current：5 to 100％ (1％ step)
Voltage：1.0 to 20.0％(0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF [Non-use] (Current, Voltage)◆ 142H Current 5th conversion content upper limit,
146H Voltage 5th conversion content upper limit.
Set the high-alarm value of 5th conversion content (current, voltage).
Setting range. Current：5 to 100％ (1％ step)
Voltage：1.0 to 20.0％ (0.1％ step)
Current setting
Setting No.
Selection by ＋ and － , set value is updated by SET .
Current distortion-factor upper limit
Default setting：OFF [Non-use] (Current, Voltage)◆ 143 Current nth content factor, 147 Voltage nth content factor.

Setting factor

New setting

Set the factor (order) of nth content (current, voltage).
An order can be selected from n= 3, 4, 5, 7, 9, 11, 13, 15.
Selection by ＋ and － , set value is updated by SET .
Default setting：5th(Current, Voltage)◆ 144H Current nth content upper limit,
148H Voltage nth content upper limit.
Set the high-alarm value of nth content (current, voltage).
Setting range. Current：5 to 100％ (1％ step)
Voltage：1.0 to 20.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF [Non-use] (Current, Voltage)

Setting No.

Current setting

Voltage nth content factor

46

SQLC-215-133◆ 149 5th conversion detection characteristics.
Alarm output, Inverse-time-delay characteristics
In case of high-limit-setting value 35％(voltmeter 3.5％)

The detection characteristics of 5th conversion content canbe selected from A (average mode) and I (inverse-time-delaymode).
In case of "A (average mode)", when average measured value(the average of instantaneous value in average timeinterval) exceeded the upper limit alarm value, it detects.
And in case of "I (inverse-time-delay mode)", when exceeding
the upper limit warning value by anti-time limit
characteristic of instantaneous value, it detects.
Selection by ＋ and － , set value is updated by SET .
Default setting：I(inverse-time-delay mode)

10000

1000

＋

100

A：Average＋

I：inverse-time-

mode

－

delay mode

Time
(sec）

－

10

New setting
Setting factor

1
0(Ammeter)(Voltmeter) 0

Setting No.

20
2

40
4

60
6

80 100 120 140 160 180 200
8 10 12 14 16 18 20

5th conversion content (％)

Current setting

5th conversion detection◆ 14A Average time interval
Set the average time interval of each harmonic measurement.
Selection by ＋ and － , set value is updated by SET .
In addition, in case the detection characteristics of 5th conversion content are set as inverse-time-delay mode,in order that 5th conversion content may operate in inverse-time-delay characteristics, this time interval isdisregarded. Default setting：0 minute (With no average)
0 minute(Nothing)＋－

1 minute＋－

2 minute＋－

5 minute＋－

10 minute＋－15 minute＋－30 minute－＋(5) 151H to 152L Instant measurement detection setting 【Current input is excluded.】
An upper limit low-alarm value is set by instant measurement (voltage factor).
151H
Instant measurement
voltage upper limit

RESET/SHIFT

152L
Instant measurement
voltage lower limit

New setting

Current setting

Setting No.

RESET/
SHIFT◆ 151H Instant measurement voltage upper limit,
152L Instant measurement voltage lower limit.
Set the high-alarm value and low-alarm value of instantmeasurement (voltage).
Setting range is 30 to 150％ (1％ step).
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF〔Non-use〕(Upper limit, lower limit )47

Instant measurement
voltage upper limit

SQLC-215-133(6) 171 to 172 Backlight setting
Sets the action and brightness of backlight.
171
Backlight action

RESET/
SHIFT

172
Backlight brightness

RESET/SHIFT

New setting

Current setting◆ 171 Backlight action
It can select from ON (always-on), AUTO (auto off), and OFF (always-off)about action of backlight.
If 5 minutes elapses without operating a switch in case it is set as "AUTO(auto off)", backlight will go out automatically.
After that, backlight will be turned on if either of switches is operated.
Selection by ＋ and － , set value is updated by SET .
Default setting：AUTO (Auto off)
AUTO：
Auto off

＋

OFF：

＋

ON：

－

Always-off－

Always-on
Setting No
Setting factor
Backlight－＋◆ 172 Backlight brightness
It can select the brightness of backlight as five steps of 1 to 5.
Backlight becomes the darkest if it is set as "1".
Backlight becomes the brightest if it is set as "5".
Selection by ＋ and － , set value is updated by SET .
Default setting：3 (Middle)

48

Setting54321Brightness
BrightDark

SQLC-215-133

5.3.2 Setting mode 2
SET andRESET/SHIFT
3 seconds
211 to 218
Measurement-range setting

DISPLAY

MODE
221A to 228A
Analog output setting

DISPLAY

MODE
241P to 244P
Pulse output setting

DISPLAY

Display mode
MODE

(Measurement
display)

251 to 252
External operation input setting

DISPLAY

MODE
261 to 26C
Measurement ON/OFF setting

DISPLAYMODE
DISPLAY

271 to 272

Initialization of watt-hour and setting value.
MODE

Setting mode 2

Setting mode 2 is selected by pressing SET and RESET/SHIFT switches continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, a display and output of measurement are not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

(1) 211 to 218 Measurement-range setting
Setting of measurement range in each measurement factors.
211
Voltage range
RESET/SHIFT

212
RESET/SHIFT

218
Frequency range

Current range

RESET/SHIFT

217
RESET/SHIFTPower-factor
range

213
Current display
intrinsic
sensitivity

214
RESET/SHIFT216
RESET/SHIFT

49

Reactive power
range

Power polarity

215
RESET/SHIFTPower range

RESET/SHIFTSQLC-215-133◆ 211 Voltage range
Set the voltage range (VT ratio). Change of this setting also sets the measurement range of active power andreactive power automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：6600V
Voltage measurement range (34 range)
Setting factor
150.0V(110V)
150V (110V)
300.0V(220V)
300V (220V)
500V (380V)
600V (440V)
600V (460V)
600V (480V)
1200V (880V)
1500V (1100V)
2400V (1650V)
3000V (2200V)
3.00kV (2200V)

4500V (3300V)
4.50kV (3300V)
9000V (6600V)
9.00kV (6600V)
15.00kV (11kV)
18.00kV (13.2kV)
18.00kV (13.8kV)
24.00kV (16.5kV)
25.00kV (18.4kV)
30.0 kV (22kV)
45.0 kV (33kV)
90.0 kV (66kV)
120.0 kV (77kV)

New setting

150.0kV (110kV)
180.0kV (132kV)
210.0kV (154kV)
270.0kV (187kV)
300.0kV (220kV)
400.0kV (275kV)
500.0kV (380kV)
750.0kV (550kV)

Current setting
Setting No.
Voltage range◆ 212 Current range
Set the current range (CT ratio). Change of this setting also sets the measurement range of active power andreactive power automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：100.0A
Current-measurement range (76 range)

5.00A
6.00A
7.50A
8.00A
10.00A
10.0A
12.00A
12.0A
15.00A
15.0A

20.00A
20.0A
25.00A
25.0A
30.00A
30.0A
40.0A
50.0A
60.0A
75.0A

80.0A
100.0A
100A
120.0A
120A
150.0A
150A
200.0A
200A
250.0A

250A
300.0A
300A
400A
500A
600A
750A
800A
900A
1000A

1.00kA
1200A
1.20kA
1500A
1.50kA
1600A
1.60kA
1800A
1.80kA
2000A

2.00kA
2500A
2.50kA
3000A
3.00kA
4000A
4.00kA
5000A
5.00kA
6000A

6.00kA
7500A
7.50kA
8000A
8.00kA
9.00kA
10.00kA
10.0kA
12.00kA
12.0kA

15.00kA
15.0kA
20.00kA
20.0kA
30.00kA
30.0kA◆ 213 Current display intrinsic sensitivity
Set the full scale of current meter.
The setting range is 40 to 120％ of CT ratio. And, it can select from the following values.
Selection by ＋ and － , set value is updated by SET .
Setting factor
Default setting：100.0A
The measurement range (current, active
power, reactive power) which can be set.
(×10n)
1.□
1.0／1.2／1.4／1.5／1.6／1.8
2.□
2.0／2.4／2.5／2.8
3.□
3.0／3.2／3.6
4.□
4.0／4.2／4.5／4.8
5.□
5.0／5.66.□
6.0／6.4
7.□
7.2／7.58.□
8.0／8.49.□
9.0／9.6

Example )In case of CT ratio＝100.0A.
･40％ of 100A is 40A.
･120％ of 100A is 120A.
A measurement range can be selected
within the limitsof 40 to 120A.
Therefore, a left table
A measurement range can be selectedfrom 40/42/45/48/50/56/60/64/72/
75/80/84/90/96/100/120A.

Setting No.

New setting

Current setting

Current display intrinsic
sensitivity

50

SQLC-215-133◆ 214 Active power polarity
A swing display of active power meter can be selected from P (one side swing) and － (both swings).
Selection by ＋ and － , set value is updated by SET .
Default setting：P (One-side swing)
New setting＋P：
One-side swing

Both swing display＋－：Both swings－－

Current setting
Setting No.

One-side swing display◆ 215 Active power range
Set the full scale of active power meter.
The setting range is 40 to 115％ of range of a VT ratio × CT ratio.
And it can select from "213 current display intrinsic sensitivity" tables.
Selection by ＋ and － , set value is updated by SET .

Power polarity<Caution> VT ratio：In case of 220V direct input. Calculates by VT ratio＝2.
CT ratio：In case of 1A input. Calculates by CT ratio÷5.
Setting factor
Default setting：1200kW

New setting◆ 216 Reactive power range
Set the full scale of reactive power meter.
The setting range is 30 to 115％ of range of a VT ratio × CT ratio.
And it can select from "213 current display intrinsic sensitivity" tables.
Selection by ＋ and － , set value is updated by SET .
<Caution> VT ratio：In case of 220V direct input. Calculates by VT
ratio＝2.
CT ratio：In case of 1A input. Calculates by CT ratio÷5.
Default setting：600kvar

Setting No.

Current setting

Reactive power range◆ 217 Power-factor range
A power-factor measurement range can be selected from 0.5‐1‐0.5/0‐1‐0.
Change of this setting also sets the analog output range of a power-factor automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：0.5‐1‐0.5＋＋

0.5‐1‐0.5

0‐1‐0

－
－

◆ 218 Frequency range
A frequency measurement range can be selected from 45.0‐55.0Hz /55.0‐65.0Hz / 45.0‐65.0Hz.
Change of this setting also sets the analog output range of afrequency automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：45.0‐65.0Hz
50：
45.0‐55.0Hz

＋
－

60：
55.0‐65.0Hz＋－

Setting factor

New setting

55：
45.0‐65.0Hz－＋
Setting No.
Current setting
Frequency range

51

SQLC-215-133(2) 221A to 228A Analog output setting 【With an analog output option】
Various setting of analog output is performed.
221A
Ao1 output
factorRESET/SHIFT

222A
Ao2 output
factorRESET/SHIFT

227A
Reactive
power output
intrinsic
sensitivity

RESET/SHIFT
228A
Low input cut

RESET/SHIFT

223A
Ao3 output
factorRESET/SHIFT

226A
Power-output
intrinsic
sensitivity

RESET/SHIFT

224A
Ao4 output
factorRESET/SHIFT

225A
Current
outputintrinsic
sensitivity

New setting
Setting No.

RESET/
SHIFT◆ 221A to 224A AO(analog output)1 to 4 output factor.
Set the output factor of each analog outputs.
Selection by ＋ and － , set value is updated by SET .
Default setting：AO1 A(L2) ，AO2 V(L1L2) ，AO3 W ，AO4 cosφ

OFF：Nothing－＋

9：DA(L3)

10：W－＋

20：Current
harmonic
5th content

21：Current
harmonic
7th－＋

30：Current
harmonic 5th
effective value
31：Current
harmonic 7th
effective value－＋

40：Voltage
harmonic 5thcontent
41：Voltage
harmonic 7thcontent－＋

50：Voltage
harmonic 5th
effective value
51：Voltage
harmonic 7th
effective value

＋
－
－
＋
＋
－

－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋

＋
－

1：V(L1L2)

8：DA(L2)

11：DW
19：Current
harmonic
4th content

22：Current
harmonic
9th
29：Current
harmonic 4th
effective value
32：Current
harmonic 9th
effective value

39：Voltage
harmonic 4thcontent
42：Voltage
harmonic 9thcontent
49：Voltage
harmonic 4th
effective value
52：Voltage
harmonic 9th
effective value

＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋

＋
－

2：V(L2L3)

7：DA(L1)

12：var
18：Current
harmonic
3th content

23：Current
harmonic 11th
content
28：Current
harmonic 3th
effective value
33：Current
harmonic 11th
effective value

38：Voltage
harmonic 3thcontent
43：Voltage
harmonic 11th
content
48：Voltage
harmonic 3th
effective value
53：Voltage
harmonic 11th
effective value

52

Current setting

Ao1 output factor

＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋

＋
－

3：V(L3L1)

6：A(L3)

13：cosφ
17：Current
harmonic 5th
conversion content

24：Current
harmonic 13thcontent
27：Current
fundamental-wave
effective value

34：current
harmonic 13th
effective value
37：Voltage
harmonic 5th
conversion content

44：Voltage
harmonic 13thcontent
47：Voltage
fundamental-wave
effective value

54：Voltage
harmonic 13th
effective value

＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋

＋
－

4：A(L1)＋－

5：A(L2)

14：Hz

＋
－

16：Current
distortion
factor
25：Current
harmonic 15thcontent＋－

26：Current harmonic
5th conversion
effective value

35：Current
harmonic 15th
effective value＋－

36：Voltage
distortion
factor
45：current
harmonic 15thcontent＋－

46：Voltage harmonic
5th conversion
effective value

55：Voltage
harmonic 15th
effective value＋－

SQLC-215-133
New setting
Current setting
Setting No.

◆ 225A Current output intrinsic sensitivity,
226A Active power output intrinsic sensitivity,
227A Reactive power output intrinsic sensitivity.
Output intrinsic sensitivity (％ of an output to an input) is set
about each analog output of current, active power, and reactive power.
The setting range can be selected from the following.
Current：40.0 to 120.0％ (0.1％ step)
Active power ：40.0 to 115.0％ (0.1％ step)
Reactive power：30.0 to 115.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：100.0％ (Current, Active power)，
50.0％ (Reactive power)

Power-output intrinsic sensitivity
Example) 9000V, 100.0A, 1200kW range, Analog output DC4‐20mA･Set 100.0％. Analog output is DC20mA in 1200kW of primary inputs.
･Set 80.0％. Analog output is DC20mA in 1200kW×0.8＝960kW of primary input.
◆ 228A Low input cut
It is the function which makes analog output a lower limit at the caseof a minute input (input which corresponds to 0.5％ or less).
A function can be selected from ON (Use) and OFF (Non-use).
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF (Non-use)
New setting
Setting No.
Setting factor
Current setting
Low input cut(3) 241P to 244P Pulse output setting 【With a pulse output option】
Various setting of a pulse output is performed.
241P
Po1 output factor

RESET/SHIFT

242P
Po1 pulse unit

RESET/SHIFT

243P
Po2 output factor

RESET/SHIFT

244P
Po2 pulse unit

RESET/SHIFTNew setting◆ 241P PO(pulse output) 1 output factor,
243P PO(pulse output) 2 output factor

Current setting

Setting No.

Set the output factor of each pulse output.
Selection by ＋ and － , set value is updated by SET .
Default setting：Wh (PO1，PO2)
OFF：
Nothing

＋

1：Wh

－

＋

2：-Wh

－

＋

3：varh
(LAG)

－

－
－
＋

6：-varh(LEAD)－＋

5：-varh
(LAG)

－

＋

4：varh(LEAD)＋

Po2 output factor
New setting◆ 242P PO(pulse output) 1 pulse unit,
244P PO(pulse output) 2 pulse unit

Current setting

Setting No.
Set the pulse unit of each pulse output.
A pulse unit can be selected from four types.
The pulse unit that can be selected is decided with full-load power.
Default setting：10kWh/p (PO1，PO2),

Po1 pulse unit

53

SQLC-215-133(4) 251 to 252 External operation input setting 【With an external operation input option】
Various setting of external operation input is performed.
251
External operation
input 1 function

RESET/SHIFT

252
External operation
input 2 function

Setting factor

New setting
ALARM and RESET aredisplayed by turns.
(Alarm reset)

RESET/SHIFT◆ 251 External operation input 1 function,
252 External operation input 2 function

Measurement factors (W,cosφ, Hz, etc.) and
DISP are displayed byturns. (Measurement
factor change)

The function of each external operation input(alarm reset, maximum / minimum reset,
measurement factor change, phase change) canbe selected.
Selection by ＋ and － , set value is
Setting No.

updated by SET .

Current setting

Default setting (With an alarm-output option)
Alarm reset (External operation input 1)
Maximum / minimum reset (External operation input 2)
Default setting (With no alarm-output option)
Maximum / minimum reset (External operation input 1)
Measurement factor change (External operation input 2)
ALARM RESET:
Alarm reset＋－

MAX./MIN. RESET：
Maximum / minimum reset＋－

Measurement factor DISP：
Measurement factor change＋－

L1,L2,L3 DISP：
Phase change－＋

・About the setting display in an external operation input function
The contents of a display at the case of function "Current setting"
Functionsetting
display point"ALARM" and "RESET" are displayed by turns by 7
Alarm reset
segment displays of the main monitor.
"RESET" is displayed by 7 segment displays of theMaximum / minimum
main monitor. And, "MAX" and "MIN" display byresetturns.
Sub-monitor"DISP" is displayed by 7 segment displays of the(right)Measurement
main monitor. And, each measurement factor (units
factor change
of A, V, W, etc.) displays by turns.
"DISP" is displayed by 7 segment displays of thePhase change
main monitor. And, each phase (L1, L2, L3, N)
displays by turns.

54"New setting"display point

Main monitor

SQLC-215-133(5) 261 to 26C Measurement ON/OFF setting
Measurement display ON/OFF setting of each measurement factor is performed.
Selection by ＋ and － , set value is updated by SET .
Default setting：ON (All measurement factors)
261
Voltage

262

RESET/SHIFTRESET/SHIFTRESET/SHIFT

Current

267
Power-receiving
watt-hourRESET/
SHIFT

268
Power
transmission
watt-hourRESET/SHIFTRESET/
SHIFT

263
Power
266
Frequency

RESET/SHIFTRESET/SHIFT

269
Power-receiving
reactive watt-hour(LAG,LEAD)
26C
Harmonic
voltage

264
Reactive
power
265

New setting
Setting No.
Current setting

RESET/SHIFTPower-factor
26A
Power transmission
reactive watt-hour(LAG,LEAD)RESET/SHIFT

26B
Harmonic
current

RESET/SHIFTRESET/
SHIFT

Voltage ON/OFF(6) 271 to 272 Initialization of watt-hour and setting value.
Package reset of watt-hour is performed. And, each set value is initialized (returns to default setting).
271
Setting value isinitialized.

RESET/SHIFT

272
Watt-hour reset

RESET/
SHIFT◆ 271 Initialization of setting value
Each set value is initialized (returns to default setting).
Pushing SET for 3 seconds initializes all set values.

SET
3 seconds

Setting No.

Setting No.

Completion of
initializing(Before initialization)(Initialization completion)

Initialization of setting value◆ 272 Watt-hour reset
Integrated value of each electric-energy display is cleared (＝0). By pushing SET for 3 seconds, all integrated
value (Wh, -Wh, var(LAG), -var(LAG), var(LEAD), var(LEAD)) is cleared by package.
Setting No.

Setting No.
Wh and varh aredisplayed byturns.

Wh and varh aredisplayed byturns.

SET
3 seconds
Clear completion
After clear
completion(Before clear)(Clear completion)
Watt-hour reset

55

SQLC-215-133

5.3.3 Setting mode 3
SET and DISPLAY
3 seconds
311 to 312
Input circuit setting

DISPLAYMODE
321 to 322
Measurement setting

DISPLAY

Display mode
MODE

(Measurement
display)

331
Analog output specification change

DISPLAY

MODE
341 to 348
Analog output adjust

DISPLAY

MODE

Setting mode 3

Setting mode 3 is selected by pressing SET and DISPLAY switches continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, a display and output of measurement are not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

(1) 311 to 312 Input circuit setting
Set the input circuit and phase wire and input voltage /phase-voltage full scale.
311
Input circuit andphase wire change

RESET/SHIFT

New setting
Current setting
Setting No.

312
Input voltage

RESET/
SHIFT◆ 311 Input circuit and phase wire change
2VT･3CT and 3CT product is 3-phase 3 line (3φ3W) fixation.

Input circuit and phase wire change(3φ3W fixation)◆ 312 Input voltage
Set the input voltage.
Selection by ＋ and － , set value is updated by SET .
Default setting：110V

New setting
Setting No.
Current setting

＋
＋
110V

220V
－
－

Input-voltage rating (3φ3W)
56

SQLC-215-133(2) 321 to 322 Measurement setting
Set the dead band of measurement display, and with or without of tidal current measurement.
321
Dead band of
measurement

RESET/
SHIFT

322
Tidal current
measurement

New setting
Current setting

RESET/SHIFT◆ 321 Measurement dead band
Set the dead band of measurement display.
By this setting, variation of less than this set value
is disregarded by voltage, current, active power, andreactive power measurement display.
Setting range：0.0 to 2.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：0.0％ (Nothing)◆ 322 Tidal current measurement
By measurement of reactive power and a power-factor,
0 (general measurement) or 1 (tidal current measurement
which was conscious of power transmission/power receiving)can be selected.
Selection by ＋ and － , set value is updated by SET .
Default setting：0 (General measurement)＋＋

0：Generalmeasurement－－

Setting No.
Setting factor

Dead band of measurement

New setting
Current setting
Setting No.

1：Tidal current

measurement
Tidal current measurement(General measurement)
Display

Analog output[mA] LEAD 0‐LAG 0/4‐20mA

LEAD 0‐LAG 0/LEAD 0‐LAG 0

LAG 0

20

cosφ=1

12

LEAD 04

Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]
LAG
cosφ LEAD
LEAD cosφ
1
0
1
0
0

Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]LEAD cosφ
LAG
cosφ LEAD
0
1
0
1
0

Power receiving

Power receiving

Power transmission

Power transmission(Tidal current measurement)
Display

LEAD 0‐LAG 0/LEAD 0‐LAG 0

Analog output[mA]

LAG 0

LEAD 0‐LAG 0/4‐20mA

20

cos180°=1
LEAD
0
LAG

12

cos0°=1
LEAD 04
Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]LEAD cosφ LAG,LEAD cosφ LAG
1
0
0
1
0
0

Input
φ=-90° φ=0° φ=90°φ=180°φ=270°[cosφ]LEAD cosφ LAG,LEAD cosφ LAG
1
0
0
1
0
0

Power receiving

Power receiving

Power transmission

57

Power transmission

SQLC-215-133(3) 331 Analog output specification setting 【With analog output (DC0‐5V or DC1‐5V)】
Set the specification of analog output (DC0‐5V／DC1‐5V).
Selection by ＋ and － , set value is updated by SET .
Default setting：DC1‐5V or DC0‐5V (Designation)
New setting＋

Current setting

＋

0‐5V
－

1‐5V
Setting No.

－

Setting factor

Analog output specification(4) 341 to 348 Analog output adjustment 【With analog output】
The bias and span of each analog output are adjusted.
341
Ao1 BIAS
adjustment

RESET/SHIFT

342
Ao1 SPAN
adjustment

RESET/SHIFT

343
Ao2 BIAS
adjustment

RESET/SHIFT

344
Ao2 SPAN
adjustment
RESET/SHIFTRESET/SHIFT
348
Ao4 SPAN
adjustment

RESET/SHIFT

347
Ao4 BIAS
adjustment

RESET/SHIFT

346
Ao3 SPAN
adjustment

RESET/SHIFT

345
Ao3 BIAS
adjustment◆ 341 AO (Analog output) 1 bias adjustment,
343 AO (Analog output) 2 bias adjustment,
345 AO (Analog output) 3 bias adjustment,
347 AO (Analog output) 4 bias adjustment.
The bias of each analog output is adjusted.
Setting range：±10.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
◆ 342 AO (Analog output) 1 span adjustment,
344 AO (Analog output) 2 span adjustment,
346 AO (Analog output) 3 span adjustment,
348 AO (Analog output) 4 span adjustment.

Ao1 span adjustment

The span of each analog output is adjusted.
Setting range：±10.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .

58

SQLC-215-133

6. Specification
6.1 Specification and intrinsic error.
Input circuit
AC110V，220V common use.
3-phase 3-wire
AC 5A or AC 1A 50/60Hz Designation

Measurement
item

Measurement range /Display specification

Voltage

AC150V to 750.0kV (34 range)

Current

Maximum demand (25), Demand, InstantAC5.00A to 30.0kA (76 range)

Input

Intrinsic error (21)Maximum Minimum
Analog output
measur- measurDigital
Pulse output
ement
ement
display22( )Current
input○

Maximum demand (25), Demand, Instant
200W to 1000MW (Range select)
Active power In voltage and current range.
One-side swing or both side swings canbe setting.

±1.0％

±0.5％

○

○

±1.0％

±0.5％

○

○

±1.0％

±0.5％

○

○

Note
L1L2-L2L3-L3L1 line
changeL1-L2-L3 phasechangePossible to set a
range of indicationand output apart fromthe CT ratio.
Range of an analog
output can be set as
indication
independently. (23)Range of an analog
output can be set as
indication
independently. (24)
In case input isbelow 20％ of voltage
range or below 2％ of
current range：cosφ=1. (Output is cosφ=1 equivalence)
0.0Hz in case input
is below 20％ of
voltage range.
Output is a lower
limit value. (Lower
limit value -1％：％
for output span)

LEAD,LAG 150var to 1000Mvar(Range select)
In voltage and current range.

±1.0％

±0.5％

○

○

LEAD 0.500‐1.000‐LAG 0.500
orPower factor
LEAD 0.000‐1.000‐LAG 0.000
Range select

±2.0％

±2.0％

○

○

45.0‐55.0Hz or
55.0‐65.0Hz or
45.0‐65.0Hz
Range select

±0.5％

±0.5％

○

○

±1.0％

±2.5％

○

±2.5％

±2.5％

○

±1.5％

±1.5％

○

Digital display is ％to a voltage range.

±1.5％

±1.5％

○

Digital display is ％to a current range.

±1.0％

±2.5％

○

Reactive
power

Frequency

Voltage
Distortion
factorCurrent
Harmonic ntheffective
Voltagevalue
Fundamentalwave
Current
effective
value
Voltage
Harmonic nth
content
Current

Harmonic 5th Voltage
conversion
effective
value
Current

0.0 to 20.0％(Second to 15th harmonic)
L1L2-L2L3
0.0 to 100.0％(Second to 15th harmonic)
L1-L2-L3○

AC150V to 750.0kV (34 range)
n=3,4,5,7,9,11,13,15
L1L2-L2L3
AC5.00A to 30.0kA (76 range)
n=3,4,5,7,9,11,13,15
L1-L2-L3
0.0 to 20.0％
n＝3,4,5,7,9,11,13,15
L1L2-L2L3
0.0 to 100.0％
n＝3,4,5,7,9,11,13,15
L1-L2-L3○

Digital display is ％to 100% of content.
○

AC150V to 750.0kV(34 range) L1L2-L2L3
AC5.00A to 30.0kA(76 range) L1-L2-L3

Digital display is ％to 100％ of
distortion factors.

○

59

±2.5％

±2.5％

○

±1.5％

±1.5％

○

±1.5％

±1.5％

○

Digital display is％ to a voltage
range.
Digital display is％ to a current
range.

SQLC-215-133

Measurement
item

Measurement range /Display specification

Harmonic 5th Voltage
conversion
Current
content

0.0 to 20.0％ L1L2-L2L3

Watt-hour

var-hour

0.0 to 100.0％

L1-L2-L3

Intrinsic error (21)Maximum Minimum
Analog output
measur- measurDigital
Pulse output
ement
ement
display(22)

Current
input

±1.0％

±2.5％

○

±2.5％

±2.5％

○

Power factor
1：±2.0％

Power factor
1：±2.0％

Power factor
0.5：±2.5％

Power factor
0.5：±2.5％

Power factor
0：±2.5％

Power factor
0：±2.5％

Power factor
0.87：±2.5％

Power factor
0.87：±2.5％○

Display：Integer, 5 digit.
Multiplier：Integral number times of 10.
Possible to indicate it to 3 decimal place.
Electric power is integrated. (Power
receiving， Power transmission)
Display：Integer, 5 digit.
Multiplier：Integral number times of 10.
Possible to indicate it to 3 decimal place.
Integrating reactive power of power receiving.
(LAG･LEAD)
Integrating reactive power of powertransmission. (LAG･LEAD)

Note
Digital display is％ to 100% ofcontent.
Conformity withnormal watt-hourmeter.
Setting range ofpulse output unit
(kWh/pulse) isreferred to optionspecification.
Setting range ofpulse output unit
(kvarh/pulse) isreferred to optionspecification.

Note(21) If this unit directly measures an inverter output of cycle control, SCR phase angle control or PWM control, anerror may increase due to its operation principle.
As for harmonic 5th conversion effective value and nth harmonic effective value, a display will be zero by 0.2％
or less of the measurement range. And, analog output constitutes lower limit output. At this time, a distortion
factor, harmonic 5th conversion content rate, and nth harmonic content also become 0％ (analog output is lower
limit output). As for a distortion factor, harmonic 5th conversion effective value / content, and nth harmonic
effective value / content, fundamental-wave effective value constitutes zero by 3％ or less of the measurement
range.
Note(22) Analog output and pulse output are options.
Analog output of harmonic distortion factor and harmonic content：It becomes an upper limit output from a lower
limit output, for current 0 to 100％ and voltage 0 to 20％.
Note(23) At the case of one side swing setting of bar graph. Digital meter measures reverse power to -15% full scale.
(Instantaneous power)
Note(24) Active power, reactive power display digit.
A full scale display is below 4000：4 digits display. A full scale display is 4000 or more：3 digits display.
Example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW
Please refer to an appendix table for details.
Note(25) The maximum value (maximum demand current, others) minimum value can be checked in the maximum and theminimum measurement mode.

60

SQLC-215-133

Item
Bar graphdisplay

OperatingmethodIntervalsetting

The factor
in whichdisplaysetting ispossible

OptionSpecification
Bar graph display of the main-monitor factor is done. (Watt-hour and var-hour exclude)
A display of a sub-monitor factor can also be set.
Current, Voltage ：Effective value computing type.
Demand ammeter：Arithmetic method according with bimetallic type.
Demand active power meter：Arithmetic method according with bimetallic type, or average value within the demand
time limit. (One side is selected by setting.)
Active power, Reactive power, Watt-hour, var-hour：Time-division multiplication method.
Power factor：Average value within instant measurement or the demand time limit. (One side is selectedby setting.) Calculates for active power and reactive power.
Frequency：Zero cross cycle computing type.
Harmonic：FFT computing type.
0 second / 5 seconds / 10 seconds / 20 seconds / 30 seconds / 40 seconds / 50 seconds /Demand current
1 minute / 2 minutes / 3 minutes / 4 minutes / 5 minutes / 6 minutes / 7 minutes /8 minutes / 9 minutes / 10 minutes / 15 minutes / 20 minutes / 25 minutes / 30 minutes
Demand active power(95％ time limit)
Harmonic
Average time limit：0 minute / 1 minute / 2 minutes / 5 minutes / 10 minutes /measurement
15 minutes / 30 minutes Average measurement.
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Demand active power, Reactive power, Power factor, Frequency,
Main monitor
Watt-hour(Power receiving, Power transmission), var-hour(Power receiving LAG/LEAD,
Power transmission LAG/LEAD), Distortion factor(A, V)
Sub-monitor (Left)
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Active power, Reactive power, Power factor
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Sub-monitor
Demand active power, Reactive power, var-hour(Power receiving LAG/LEAD,(Center)
Power transmission LAG/LEAD), Harmonic 5th conversion content(A, V),
Harmonic nth content(A, V)
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Demand active power, Power factor, Frequency, Watt-hour(Power receiving,Sub-monitor (Right)
Power transmission), Fundamental-wave effective value(A, V),
Harmonic 5th conversion content(A, V), Harmonic nth effective value(A, V)
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Demand active power, Reactive power, Power factor, Frequency, Distortion factor(A, V),
Bar graph
Harmonic 5th conversion content(A, V), Harmonic nth content(A, V),
Fundamental-wave effective value(A, V), Harmonic 5th conversion effective value(A, V),
Harmonic nth effective value(A, V)
Analog output (4 sets)
Contact output (2 sets. Select of pulse output or alarm output or CPU error output.)
External operation change input (2 sets)● About power and reactive power full scale range selection
The power range and reactive power range is automatically decided in a current range and voltage range.
The full scale range of a bar graph can be selected out of the following values within a 40 to 115％ (range of power)and 30 to 115％ (range of reactive power), assuming that the rated power (VT ratio × CT ratio)(26) is 100％.
1.0/1.2/1.4/1.5/1.6/1.8/2.0/2.4/2.5/2.8/3.0/3.2/3.6/4.0/4.2/4.5/4.8/5.0/5.6/6.0/6.4/7.2/7.5/8.0/8.4/9.0/9.6 ×10n
Example ) In case of VT ratio × CT ratio ＝ 1200kW
A full scale range can be selected from the following.
480/500/560/600/640/720/750/800/840/900/960/1000/1200
Note(26) Assume VT ratio is “2” for calculation in case of 220V input specifications and also assume CT ratio is
CT ratio ÷ 5 in case of 1A input specifications.

61

SQLC-215-133● Measurement is possible range.
Input (27)

Measurement factor
VoltageAC0‐150V [AC0‐300V]

Current

AC0‐5A[AC0‐1A]

0‐1kW (0‐200W)[0‐2kW (0‐400W)]
LEAD 1‐0‐LAG 1kvar(LEAD 200‐0‐LAG 200var)[LEAD 2‐0‐LAG 2kvar(LEAD 400‐0‐LAG 400var)]

Power

Reactive power

LEAD 0‐1‐LAG 0Power factor
LEAD 0.5‐1‐LAG 0.5

Current

45.0‐55.0Hz
55.0‐65.0Hz
45.0‐65.0Hz
AC0‐5A[AC0‐1A]

VoltageAC0‐150V [AC0‐300V]

Frequency
Effective
value
Harmonic
Rate ofcontent
Note(27) [Current
Voltage0 to 100％(％ to input rating)
0 to 20％(％ to meter full scale)] is the 300V input case. (Measurement is possible range
Display
Analog output
101％ of meter full
101％ of output span.
scale.
120％ of input
120％ of output span.
rating.
120％ of input-1％ and 120％ of
rating.
output span,
120％ of input
rating.

-1％ and 120％ of
output span,

LEAD 0.000‐1‐LAG 0.000
LEAD 0.490‐1‐LAG 0.490
44.9‐55.1Hz
54.9‐65.1Hz
44.8‐65.2Hz
120％ of input rating.
101％ of meter fullscale.
120％ of input
rating.
100％ of meter fullscale.

0％ and 100％ of
output span,-1％ and 101％ of
output span,

Remarks

Instant,
DemandInstant,
Demand-1％ and 101％ of
output span,
120％ of output span.
101％ of output span.
120％ of output span.
120％ of output span.

) is the 1A input case.

6.2

Specification, Performance.
Item
Specification
Accuracy
Reference to measure specification and accuracy
Accuracy of bar graph±10％ （％ for span）
Influence by temperature 23±10℃ within accuracy.
JIS C 1102-1：1997 〔IEC60051-1 1984 ，Amendment 1(1994)，Amendment 2(1995)〕
JIS C 1102-2：1997 〔IEC60051-2 1984〕
JIS C 1102-3：1997 〔IEC60051-3 1984 ，Amendment 1(1994)〕
JIS C 1102-4：1997 〔IEC60051-4 1984〕
Conformity technical
JIS C 1102-5：1997 〔IEC60051-5 1985〕
standard
JIS C 1102-7：1997 〔IEC60051-7 1984〕
JIS C 1111：1989
JIS C 1216：1995
JIS C 1263：1995
Electro-magnetic compatibility directive (EMC directive) 2004/108/EC
CE compliant directive
Low voltage directive (LVD) 73/23/EEC
JIS C 1010-1：2005 (IEC 61010-1：2001)
Measurement category Ⅲ(The category to the measurement performed with fabric equipment)
Maximum use voltage, 300V
SafetyDegree of contamination, 2(Usually, the grade which only contamination of non-conductivity causes.
However, temporary conductivity, which originates in dew condensation depending on
the case occurs.)
About 1 second (Bar graph：0.25 seconds)
Display updating time
(Harmonic measurement：Digital and bar graph is 10 seconds or less.)
Main monitor
5 digit, character height 11mm
Sub-monitor (Left)
4 digit, character height 6mm
Display device
LCD
Display composition
Sub-monitor (Center),(Right)
5 digit, character height 6mm
Bar graph
20 dots

62

SQLC-215-133

Item
LCD view angle
Backlight

Auxiliary supply

Rush current(Time constant)

Input consumption VA

Overload capacity

Insulation resistance
JIS C 1102-1
JIS C 1111

Withstand voltage
JIS C 1102-1
JIS C 1111

Impulse withstandvoltage
JIS C 1111

Noise-capacity
ANSI C37.90a
JEA B-402
IEC801-2

Specification
For upper installation
Upper view angle 10°，Lower view angle 60°，
Standard(For lower view)
Right and left view angle 60°
LED backlight：White
Always-on, Auto off (after 5 minutes without operating), Always-off. Setting is possible.
Backlight can select brightness from five steps of 1 to 5.
(1) AC85‐264V 50/60Hz 10VA (Rated voltage, AC100/110V，200/220V)
DC80‐143V
6W (Rated voltage, DC100/110V) for both AC and DC uses
(2) DC20‐56V
6W (Rated voltage, DC24/48V)
Rated voltage AC110V 2.2A or less (About 3.6ms)
Rated voltage AC220V 4.4A or less (About 3.6ms)
Rated voltage DC110V 1.6A or less (About 3.6ms)
Rated voltage DC24V
5.0A or less (About 2.0ms)
Rated voltage DC48V
9.9A or less (About 2.0ms)
Voltage circuit
0.1VA or less (110V) ，0.2VA or less (220V)
Current circuit
0.1VA or less (5A,1A)
Voltage circuit
2 times 10 seconds, 1.2 times continuation of rated voltage.
40 times 1 second, 20 times 4 seconds, 10 times 16 seconds,
Current circuit
1.2 times continuation of rated current.
1.5 times 10 seconds, 1.2 times continuation of rated voltage.
Auxiliary supply
In case of DC110V, 1.5 times 10 seconds, 1.3 times continuation of rated
voltage.
Between electric circuits and case (Earth).
Between input and output and auxiliary supply.
Between analog output and pulse output and alarm output. Above 50MΩ at DC500V megger
Between pulse outputs.
Between alarm outputs.
Between analog outputs are not insulation of minus common.
Between electric circuits and case (Earth).
AC2000V (50/60Hz) 1 minuteBetween input and output and auxiliary supply.
Between analog output and pulse output and alarm output.
AC1500V (50/60Hz) 1 minuteBetween pulse outputs.
Between alarm outputs.
Between analog outputs are not insulation of minus common.
Between electric circuits and case (Earth).
6kV 1.2/50μs Positive and negative(An analog output is excluded)polarities, for each 3 time.
5kV 1.2/50μs Positive and negative
Between analog output and case (Earth).
polarities, for each 3 time.
(1) Oscillatory surge voltage
If a vibration damping waveform (1‐1.5MHz, Peak voltage：2.5‐3kV) is repeated and added,
a measurement error should be within 10％. And, there needs to be no malfunction.
Voltage input circuit (Normal / Common),Current input circuit (Common)
Auxiliary supply circuit (Normal / Common)(2) Square wave impulse noise
If a noise (1μs, 100ns width) is repeated and added, a measurement error should be within
10％. And, there needs to be no malfunction.
Auxiliary supply circuit (Normal / Common) Over 1500V
Voltage input circuit (Normal / Common) Over 1500V
Current input circuit (Common) Over 1500V
Pulse output (Common) Over 1000V
Alarm output (Common) Over 1000V
Operation input (Common) Over 1000V
Analog output circuit (Induction) Over 1000V(3) Electric wave noise
If intermittence irradiation of the electric wave of a 150,400,900MHz band is done by (5W,1m), a measurement error should be within 10％. And, there needs to be no malfunction.
(4) Electrostatic noise
It is within 10％ of errors in 8kV electrostatic noise at the case of an energization.
There needs to be no 10kV damage at the case of a non-energization.
Condenser charge form.

63

SQLC-215-133

Item
Vibration, Shock
JIS C 1102-1
JIS C 0040，JIS C 0041

Vibration：Single amplitude 0.15mm, 10 to 55Hz, Each minute octave in 5 times sweep
Shock：490m/s２ It is each 3 times to the X,Y,Z direction.
Dimension：110mm(Width)×110mm(Height)×103.5mm(Depth) Body diameter：99mmφ
With terminal coverConstruction
International
protection code
Material
Color
MassBlackout guarantee
Operating temperatureand humidity limits
Storage temperature
limitsInstallation altitude

6.3 Option
Item

Specification

IP code：IP40
Case，Cover：ABS(V-0) ，Terminal block：PBT ，Terminal cover：Polycarbonate
Black (Munsell N1.5)
Approx. 600g
Maximum value, Minimum value, Integrates value and each setting value.
Data hold by nonvolatile memory.
-10 to +55℃, 30 to 85％ RH, Non condensing.
-25 to +70℃
The altitude of 2000m or less.

Specification
4 circuits
DC4‐20mA(Below 550Ω)
DC0‐1mA(Below 10kΩ)
OutputDC0‐5V/1‐5V (Over 600Ω)
specification
DC0‐10V(Over 2kΩ)One of them is designated by identical ratings.
Voltage(L1L2-L2L3-L3L1)，Current(L1-L2-L3)，Demand current(L1-L2-L3)，Active power，Demand active power，Reactive power，Power factor，Frequency，Distortion factor(A，V)，
Fundamental-wave effective value (Maximum value of each A/V phase)，
Output factor
Harmonic 5th conversion content (Maximum value of each A/V phase)，
Harmonic 5th conversion effective value (Maximum value of each A/V phase)，
Harmonic nth content (Maximum value of each A/V phase)，
nth harmonic effective value (Maximum value of each A/V phase)
1 second or less (Time within ±1％ of final constant value.)Response time
Harmonic measurement is 10 seconds or less.
Output ripple
Within the double precision of accuracy (％ for output span)
Between analog outputs are not insulation of minus common.
Possible output ：Watt-hour or var-hour.
Output form
：Optical MOS-FET relay. 1a contact
Contact capacity：AC，DC125V, 70mA (Resistance load, inductive load)
Pulse width：250±10ms (Output pulse width when the output pulse period of rated active power
constitutes speed more than 2 pulse/second by setting of an voltage
measurement range, a current-measurement range, and an output pulse unit is
set to 100 to 130ms.)
Setting of output pulse unit is possible by the next range.
■ 3-phase 3-wire：Full load power (kW,kvar)＝√3×Rated voltage(V)×Rated current(A)×10-3
Full load power (kW,kvar)
Output pulse unit. kWh(kvarh)/pulse
Multiplying factor
Below 1
0.1
0.01
0.001
0.00010.01 (28)Over 1
Below 10
1
0.1
0.01
0.001
0.1
Over 10
Below 100
10
1
0.1
0.01
1
Over 100
Below 1,000
100
10
1
0.1
10
Over 1,000
Below 10,000
1,000
100
10
1
100
Over 10,000 Below 100,000
10,000
1,000
100
10
1,000
Over 100,000 Below 1,000,000
100,000
10,000
1,000
100
10,000
Number of output

Analog output

Pulse output(29)

Note(28) Although multiplying factor is 0.01, a multiplying factor display will be 0.1.
(The place by the integer is 4 digits display. An enlarged display is 4 digits below decimal point.)
Note(29) 2 outputs combination is possible, for pulse output and alarm output and CPU error output.
(CPU error output is 1 point only)

64

SQLC-215-133

Item

Alarm output(30)

CPU error
output(30)

External
operation
input

Specification
Alarm factor：Demand current, Demand active power, harmonic 5th conversion content,
Harmonic nth content, Distortion factor, Voltage, Alarm OFF.
Possible to setting one of them.
Reset form ：Automatic reset or Manual reset (Setting)
Output contact ：No-voltage a contact (OR of each phase detection)
Contact capacity：AC250V 8A, DC125V 0.3A (Resistance load) AC250V 2A, DC125V 0.1A (Inductive load)
Alarm factor
Item
Specification
Demand measurement value ≧ Upper limit setting value,
Demand current, Function
Alarm display, Alarm output.
Demand active
Setting accuracy±1.0％ (％ for full scale )power
Setting range
5 to 100％ to the maximum scale. (1％ step)
Measurement value ≧ Upper limit setting value,
Function
Alarm display, Alarm output. (Detects by the maximum phase)
Setting accuracy
Current：±2.5％, Voltage：±1.0％, ％ to 100％ of content rate.
Harmonic 5th conversion content,
Harmonic nth content (n=3,4,5,7,9,11,13,15),
Harmonic 5th
Current
Distortion factor
conversion
5 to 100％ (1％ step)
content
Setting range
Harmonic 5th conversion content,
Harmonic nth
Harmonic nth content (n=3,4,5,7,9,11,13,15),
Voltage
content
Distortion factor
1.0 to 20.0％ (0.1％ step)Distortion
Average value mode：It will detect, if a par measurement value
factorbecomes beyond the above setting.
Detection
Inverse-time-delay mode：It detects in the inverse-time-delay
characteristics
characteristics of instantaneous value(only a harmonic 5th conversion content
is possible).
Measurement value ≧ Upper limit setting value,
Alarm display, Alarm output. (Detects by the maximum phase)
Function
Measurement value ≦ Lower limit setting value,
Voltage
Alarm display, Alarm output. (Detects by the minimum phase)
Setting accuracy±1.0％ (％ for the value which set a full scale 150％)
Setting range
Using a full scale as 150％. 30 to 150％ (1％ step)
Detection item (Self-diagnostics item)
Contact composition
Contact capacity(1) Watchdog timer (Inside, outside)(2) RAM check error
AC250V 5A, DC125V 0.2A
(3) A/D-conversion error(Resistance load)b contact
At the case of error detection and not applying for
AC250V 1.5A, DC125V 0.1A
auxiliary supply, an output is ON.
(Inductive load)
It becomes OR output of a detection item.
Number of circuits
2 circuits, A function (four types) is changed by setting.
Four types of following functions can be operated by adding a voltage signal
Functionfrom the outside in addition to switch operation.
Alarm reset
Alarm output is reset (output OFF).
Maximum / Minimum The maximum/minimum value is reset (it updates to the instantaneous value atvalue reset
the time).
Measurement factor
The measurement display element of the main monitoring is changed.
changePhase change
A phase/lines display of all the current/voltage currently displayed is changed.
Minimum operation
300ms ，Continuation applying is possible.
pulse width
Input rating becomes the same as that of auxiliary supply.
(1) AC100/110V 0.4VA，AC200/220V 1.4VA，DC100/110V 0.4W AC DC two ways.
Rated input
Contact capacity：About 3mA (AC,DC100/110V) ，About 6mA (AC200/220V)(2) DC24V 0.3W，DC48V 1.2W
Contact capacity : About 10mA (DC24V) ，About 20mA (DC48V)

Note(30) 2 outputs combination is possible, for pulse output and alarm output and CPU error output.
(CPU error output is 1 point only)

65

SQLC-215-133● Caution on the use of external display selection input (option)
External power consumption is 0.4VA at AC110V or 1.4VA at AC220V or 0.4W at DC110V.
In case a relay or a switch is used for power-supply supply, please use the thing of about
1mA of the minimum application loads.

P N
5
6
28
29

6.4 CE marking(1) Conformity technical standard
・EMC conformity technical standard
EMI(Emission) EN61000-6-4
EMS(Immunity) EN61000-6-2
・Safety technical standard
EN61010-1 CATⅢ(Maximum circuit voltage, 300V)，Pollution degree 2(2) Specification according to EMC test item
EMC test item

EMI

Electromagnetic
radiation
disturbanceElectrostatic
discharge
Radio-frequency
amplitude modulated
electromagnetic
field
Fast transients
Surge

Limits

Quasi-peak 10m
distance
Quasi-peak 10m
230 to 1000MHz
distance
Contact discharge±4kV (Charge voltage)
Air discharge±8kV (Charge voltage)
30 to 230MHz

80 to 1000MHz

10V/m，80％ AM(1kHz)

1.4 to 2.0GHz

3V/m，80％ AM(1kHz)

2.0 to 2.7GHz

1V/m，80％ AM(1kHz)

Power port ：Line to earth ±2.0kV
Signal port：Line to earth ±1.0kV
Power port ：Line to line ±0.5kVLine to earth ±0.5kV
Signal port：Line to earth ±1.0kV

Radio-frequency
common mode

0.15 to 80MHz

10V，80％ AM(1kHz)

Power-frequency
magnetic field

50/60Hz

30A/m

EMS

0％：1 cycle

Voltage dip andinstantaneous power
failureVoltage dip％ residual
voltage40％：10/12 cycle
70％：25/30 cycle

Instantaneous
power failure％ residual
voltage0％：250/300 cycle

6640dB(μV/m)

Conformity
technical standard

47dB(μV/m)

EN61000-6-4：2001
EN55011：1998
classA，Group1

After test：
Performance is satisfied.

EN61000-6-2：2005
EN61000-4-2：1995

During test：
Measurement error ±20％
After test：
Performance is satisfied.
After test：
Performance is satisfied.
After test：
Performance is satisfied.
During test：
Measurement error ±20％
After test：
Performance is satisfied.
During test：
Measurement error ±20％
After test：
Performance is satisfied.
After test：
Performance is satisfied.
After test：
Performance is satisfied.
(Function loss to a first orderis allowable, and thing which
can do self-recovery.)After test：
Performance is satisfied.
(Function loss to a first orderis allowable, and thing which
can do self-recovery.)EN61000-6-2：2005
EN61000-4-3：2006
EN61000-6-2：2005
EN61000-4-4：2004
EN61000-6-2：2005
EN61000-4-5：2006
EN61000-6-2：2005
EN61000-4-6：1996

EN61000-6-2：2005
EN61000-4-8：1993

EN-61000-6-2：2005
EN-61000-4-11：2004

SQLC-215-133

7. Maintenance and check7.1 Trouble shooting
Symptoms
Possible causes
The power supply is not supplied.
(Not connected. or voltage is low)
It does not display.
Measurement display ON/OFF setting is set to OFF.
Trouble of device.
Range is not set correctly.
Measuring value error Out of range in rated frequency (45‐65Hz).
is noticeable
Cycle control, SCR phase angle control, PWM, or otherinverter output is measured.
Analog output is not Analog output is set to OFF or measurement factor is
outputted.
set to OFF.
Pulse output is not
Pulse output is set to OFF or measurement factor is
outputted.
set to OFF.
Alarm output does not
The return method is a "manual reset".
return.

Remedial measures
Check the auxiliary supply.
Again, a power supply is supplied.
Setting check.
Replace the device.
Please set again.
It cannot be used.
It cannot be used.
Please check a setting
Please check a setting
Please check a setting

7.2 CheckBe careful with the following items periodically.
(1) Check if the measuring values, scale digits, units, etc. are displayed correctly.
(2) Check if LCD is free of a color change, breakage of case, or other defects.
(3) Check the unit for loose wiring and loose mounting screws.
(4) Please remove, if dust has adhered to the product.
7.3 Maintenance(1) Please do not make a wiring change of the input and auxiliary supply in an energization.
(2) In case you check an output in an energization, please warn to be unable to touch output wiring and a human body to
an input and an auxiliary supply terminal. (Please do not short-circuit a voltage output.)(3) Please check a connection diagram, in case you check the voltage of input, output, and auxiliary supply.
(4) If a name-plate is wiped with solvents (an alcoholic nature), a display item may disappear. Please wipe off withthe dry cloth.

67

SQLC-215-133

7.4 Test

In case this product is tested, a special setting or operation is not needed fundamentally.
test should operate it along with each process.

However, the following(1) Alarm-output test
Even if this product does not have input, it can perform ON/OFF test of an alarm output (relay-contact output).
Operation is performed by alarm 1 test and alarm 2 test in the setting mode 1.
Please refer to "5.3.1 Setting mode 1 (2) alarm-output setup" about the details of operation.
In addition, an alarm return constitutes action corresponding to a return method setting (automatic reset / manual
reset) of each alarm output.
< Operating process >・Alarm 1 test (Setting № 124)
(111)
(121)
(124)
Press SET for 3 seconds → Press MODE → Press RESET/SHIFT for 3 times → While pushing SET , thealarm 1 output is ON.
・Alarm 2 test (Setting № 128)
(111)
(121)
(128)
Press SET for 3 seconds → Press MODE → Press RESET/SHIFT for 7 times → While pushing SET , thealarm 2 output is ON.
(2) Harmonic time-interval test
Please test by the following operation about the time interval of a harmonic. An error becomes large, in case it
tests without performing the following operation.
Operation is performed with the average time interval in the setting mode 1.
Please refer to "5.3.1 setting mode 1 (4) harmonic detection setting" for operation details.
< Operating process > (Setting № 14A)
(111)
(141)
(14A)
Press SET for 3 seconds → Press MODE for 3 times → Press RESET/SHIFT for 9 times
Please press SET simultaneously with a test start.
Please press DISPLAY and return to a measurement screen.
7.5 Storage(1) Please avoid storage in the next space. Low temperature, high temperature, high humidity, and sunny place.
(2) The aluminum electrolytic capacitor is used for a product. Please do the energization of the power supply within
one year after shipment.
7.6 Countermeasures against troubles
As our principle, we recall product in question and repair it. If judged as product failure, have a contact with us
or sales agent for repairing work (Also have a contact with us or sales agent for specification change).
Product failure which we are not responsible for (When responsibility in manufacturing process is not recognized, when
product is disassembled/remodeled, in case of false operation by customer, etc.) is beyond our warranty.

68

SQLC-215-133
Appendix table 1-1
Active power range, watt-hour multiplier rate table (3-phase 3-wire)
V range
Multiplierrate
A range
5A[W]
5.00 M

6A

6.00 M

7.5A

×100

8.00 M

10A

10.00 M

12A

12.00 M

15A

15.00 M

20A

20.00 M

25A

25.00 M

30A

30.00 M

40A

40.0 M

60A
75A
80A
100A
120A
150A

500.0kV

50.0 M
60.0 M
75.0 M
80.0 M
100.0 M
120.0 M
150.0 M

375.0kV

300.0kV[W]
2500 k[W]
2000 k

3000 k2400 k

4.00 M(3.75)
4.00 M

3000 k

5.00 M

4.00 M

6.00 M

4.80 M

3200 k

7.50 M

6.00 M

10.00 M

8.00 M

14.00 M
(12.50)15.00 M

10.00 M

20.00 M

16.00 M

25.00 M

20.00 M

30.00 M

24.00 M

40.0 M(37.5)
40.0 M

30.00 M
32.00 M

50.0 M

40.0 M

60.0 M

48.0 M

75.0 M

60.0 M

12.00 M

80.0 M

140.0 M
(125.0)150.0 M

100.0 M

200.0 M

160.0 M

250.0 M

200.0 M

300.0 M

240.0 M

400 M
(375)
400 M

300.0 M
320.0 M

450 M

360.0 M

500 M

400 M

600 M

480 M

1500A

750 M

600 M

1600A

800 M

640 M

900 M

720 M

1000 M

800 M

250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A

1800A
2000A
2500A
3000A

200.0 M[W]
3600 k(3455)
4.20 M(4.15)
5.60 M(5.18)
5.60 M(5.53)
7.20 M(6.91)
8.40 M(8.29)
10.00 M
(10.36)14.00 M
(13.82)18.00 M
(17.27)20.00 M
(20.73)28.00 M
(27.64)36.00 M
(34.55)42.0 M(41.5)
56.0 M(51.8)
56.0 M(55.3)
72.0 M(69.1)
84.0 M(82.9)
100.0 M
(103.6)140.0 M(138.2)180.0 M
(172.7)200.0 M
(207.3)280.0 M
(276.4)360.0 M
(345.5)
420 M
(415)
560 M
(518)
560 M
(553)
640 M
(622)
720 M
(691)
840 M
(829)

100.0 M

200A

×100007.50 M

8A

50A

×1000

750.0kV

255.0kV

210.0kV

180.0kV

150.0kV

105.0kV

90.0kV

45.0kV

30.0kV(VT550000/110V) (VT380000/110V) (VT275000/110V) (VT220000/110V) (VT187000/110V) (VT154000/110V) (VT132000/110V) (VT110000/110V) (VT77000/110V) (VT66000/110V) (VT33000/110V) (VT22000/110V) Multiplier

250.0 M
300.0 M
400 M
500 M
600 M
750 M
800 M
900 M
1000 M

120.0 M

1000 M

[W]
1800 k(1700)
2000 k(2040)
2500 k(2550)
3000 k(2720)
3600 k(3400)
4.20 M(4.08)
5.60 M(5.10)
7.20 M(6.80)
9.00 M(8.50)
10.00 M
(10.20)14.00 M
(13.60)18.00 M
(17.00)20.00 M
(20.40)28.00 M
(25.50)28.00 M
(27.20)36.00 M
(34.00)42.0 M(40.8)
56.0 M(51.0)
72.0 M(68.0)
90.0 M(85.0)
100.0 M
(102.0)140.0 M
(136.0)180.0 M
(170.0)200.0 M
(204.0)280.0 M
(255.0)280.0 M
(272.0)320.0 M
(306.0)360.0 M
(340.0)
420 M
(408)
560 M
(510)
560 M
(544)
640 M
(612)
720 M
(680)
900 M
(850)

[W]
1400 k[W]
1200 k[W]
1000 k1800 k(1680)
2400 k(2100)
2400 k(2240)
2800 k1500 k(1440)
1800 k1200 k2000 k(1920)
2400 k1600 k3600 k(3360)
4.20 M

3000 k(2880)
3600 k2400 k

5.60 M

4.80 M

4.00 M

7.20 M(7.00)
8.40 M

6.00 M

5.00 M

7.20 M

12.00 M
(11.20)14.00 M

[W]
600 k

[W]
300.0 k[W]
200.0 k720 k360.0 k240.0 k

900 k

450 k

300.0 k

960 k

480 k

320.0 k1200 k

600 k

400 k

1800 k(1680)
2400 k(2100)
2800 k1500 k(1440)
1800 k

720 k

480 k

900 k

600 k

2400 k1200 k800 k3600 k(3500)
4.20 M

3000 k1500 k1000 k

6.00 M

3600 k1800 k1200 k1500 k2000 k

3000 k

9.60 M

8.00 M

5.60 M

4.80 M

2400 k1600 k

10.00 M

6.00 M

3000 k2000 k12.00 M

7.20 M(7.00)
8.40 M

7.20 M

3600 k2400 k12.00 M
(10.50)12.00 M
(11.20)14.00 M

9.00 M

4.50 M

3000 k

15.00 M
(14.40)18.00 M

15.00 M

20.00 M
(19.20)24.00 M

20.00 M

36.00 M
(33.60)42.0 M

30.00 M
(28.80)36.00 M

24.00 M

56.0 M

48.0 M

40.0 M

72.0 M(70.0)
84.0 M

60.0 M

50.0 M

72.0 M

120.0 M
(112.0)140.0 M

16.00 M

9.60 M

4.80 M

3200 k12.00 M

6.00 M

4.00 M

18.00 M
(16.80)24.00 M
(21.00)28.00 M

15.00 M
(14.40)18.00 M

7.20 M

4.80 M

9.00 M

6.00 M

24.00 M

12.00 M

8.00 M

30.00 M

15.00 M

10.00 M

60.0 M

36.00 M
(35.00)42.0 M

36.00 M

18.00 M

12.00 M

96.0 M

80.0 M

56.0 M

48.0 M

24.00 M

16.00 M

120.0 M

100.0 M

72.0 M(70.0)
84.0 M

60.0 M

30.00 M

20.00 M

72.0 M

36.00 M

24.00 M

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M

90.0 M

45.0 M

30.00 M

96.0 M

48.0 M

32.00 M

120.0 M(108.0)120.0 M

56.0 M(54.0)
60.0 M

36.00 M

180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

150.0 M
(144.0)180.0 M

72.0 M

48.0 M

90.0 M

60.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

96.0 M

64.0 M

120.0 M(108.0)120.0 M

72.0 M

360.0 M
(350.0)420 M

300.0 M

150.0 M

100.0 M

360.0 M

180.0 M

120.0 M

30.00 M

180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

150.0 M
(144.0)180.0 M

120.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

160.0 M

360.0 M
(336.0)420 M

300.0 M
(288.0)360.0 M

240.0 M

450 M
(448)
560 M
(504)
560 M

400 M
(384)
450 M
(432)
480 M

320.0 M

720 M
(700)
840 M

600 M

500 M

720 M

600 M

960 M

5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A

150.0 M

180.0 M
200.0 M

300.0 M

360.0 M
400 M

rate

×10

12.00 M

18.00 M
(16.80)24.00 M
(21.00)24.00 M
(22.40)28.00 M

4000A

[W]
720 k
(700)
900 k
(840)
1200 k(1050)
1200 k(1120)
1400 k×100

40.0 M

80.0 M

×1000

800 M

560 M

480 M

240.0 M

160.0 M

1000 M

720 M
(700)
840 M

600 M

300.0 M

200.0 M

720 M

360.0 M

240.0 M

900 M

450 M

300.0 M

960 M

480 M

320.0 M

560 M
(540)
600 M

360.0 M

720 M

480 M

900 M

600 M

400 M

800 M

20000A

×1000030000A<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW69<Note 2>
In case the voltage range and the current range
were set asof an upper table and an
output pulse unit (a four-step setup ispossible) is set as the fastest. The output
pulse width of a pulse output is set to 100 to
130ms. (Usually 240 to 260ms)

SQLC-215-133
Appendix table 1-2
Active power range, watt-hour multiplier rate table (3-phase 3-wire)
V range
Multiplierrate
A range
5A
6A
7.5A
8A
10A
12A
15A
20A
25A

×10

30A
40A
50A
60A
75A
80A
100A
120A
150A
200A
×100

250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A

×1000

2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A

×1000020000A
30000A

25.00kV

24.00kV

18.00kV

18.00kV

15.00kV

9000V

4500V

3000V

2400V

1500V

1200V

600V

(VT18400/110V) (VT16500/110V) (VT13800/110V) (VT13200/110V) (VT11000/110V) (VT6600/110V) (VT3300/110V) (VT2200/110V) (VT1650/110V) (VT1100/110V) (VT880/110V) (VT480/110V) Multiplier[W]
180.0 k(167)200.0 k(201)
280.0 k(251)
280.0 k(268)
360.0 k
(335)
420 k
(401)
560 k
(502)
720 k
(669)
840 k
(836)
1000 k(1004)
1400 k(1338)
1800 k(1673)
2000 k(2007)
2800 k(2509)
2800 k(2676)
3600 k(3345)
4.20 M(4.01)
5.60 M(5.02)
7.20 M(6.69)
8.40 M(8.36)
10.00 M
(10.04)14.00 M
(13.38)18.00 M
(16.73)20.00 M
(20.07)28.00 M
(25.09)28.00 M
(26.76)32.00 M
(30.11)36.00 M
(33.45)42.0 M(40.1)
56.0 M(50.2)
56.0 M(53.5)
64.0 M(60.2)
72.0 M(66.9)
84.0 M(83.6)
100.0 M
(100.4)140.0 M
(133.8)180.0 M(167.3)200.0 M
(200.7)280.0 M
(250.9)280.0 M
(267.6)320.0 M
(301.1)360.0 M
(334.5)
420 M
(401)
560 M
(502)
720 M
(669)

[W]
150.0 k180.0 k240.0 k(225.0)240.0 k300.0 k360.0 k
450 k
600 k
750 k
900 k
1200 k1500 k1800 k2400 k(2250)
2400 k
3000 k3600 k
4.50 M
6.00 M
7.50 M
9.00 M
12.00 M
15.00 M
18.00 M
24.00 M
(22.50)24.00 M
28.00 M
(27.00)30.00 M
36.00 M
45.0 M
48.0 M
56.0 M(54.0)
60.0 M
75.0 M
90.0 M
120.0 M
150.0 M
180.0 M
240.0 M
(225.0)240.0 M
280.0 M
(270.0)300.0 M
360.0 M
450 M
600 M
900 M

[W]
140.0 k(125.5)160.0 k(150.5)200.0 k(188.2)200.0 k(200.7)280.0 k(250.9)320.0 k(301.1)
400 k
(376)
560 k
(502)
640 k
(627)
800 k
(753)
1000 k(1004)
1400 k(1255)
1600 k(1505)
2000 k(1882)
2000 k(2007)
2800 k(2509)
3200 k(3011)
4.00 M(3.76)
5.60 M(5.02)
6.40 M(6.27)
8.00 M(7.53)
10.00 M
(10.04)14.00 M
(12.55)16.00 M
(15.05)20.00 M(18.82)20.00 M
(20.07)24.00 M
(22.58)28.00 M
(25.09)32.00 M
(30.11)40.0 M(37.6)
42.0 M(40.1)
48.0 M(45.2)
56.0 M(50.2)
64.0 M(62.7)
80.0 M(75.3)
100.0 M
(100.4)140.0 M
(125.5)160.0 M
(150.5)200.0 M
(188.2)200.0 M
(200.7)240.0 M
(225.8)280.0 M
(250.9)320.0 M
(301.1)
400 M
(376)
560 M
(502)
800 M
(753)

[W]
120.0 k[W]
100.0 k[W]
60.0 k[W]
30.00 k[W]
20.00 k[W]
15.00 k[W]
10.00 k[W]
8.00 k

150.0 k(144.0)180.0 k120.0 k72.0 k36.00 k24.00 k18.00 k12.00 k9.60 k

150.0 k90.0 k

45.0 k30.00 k12.00 k160.0 k96.0 k

48.0 k

32.00 k24.00 k(22.50)24.00 k15.00 k200.0 k(192.0)240.0 k16.00 k200.0 k120.0 k60.0 k

40.0 k30.00 k20.00 k14.00 k(12.80)16.00 k300.0 k(288.0)360.0 k240.0 k72.0 k

48.0 k36.00 k24.00 k300.0 k150.0 k(144.0)180.0 k90.0 k

60.0 k

45.0 k30.00 k20.00 k(19.20)24.00 k

480 k

400 k

240.0 k120.0 k80.0 k

60.0 k

40.0 k

32.00 k

600 k

500 k

300.0 k150.0 k100.0 k75.0 k

50.0 k

40.0 k

720 k

600 k

360.0 k180.0 k120.0 k90.0 k

60.0 k

48.0 k

960 k

800 k

480 k

240.0 k160.0 k120.0 k80.0 k

64.0 k1200 k1000 k600 k300.0 k200.0 k150.0 k100.0 k80.0 k1500 k(1440)
1800 k1200 k720 k360.0 k240.0 k180.0 k120.0 k96.0 k1500 k

900 k

450 k

300.0 k150.0 k120.0 k160.0 k140.0 k(128.0)160.0 k2000 k(1920)
2400 k1600 k

960 k

480 k

320.0 k240.0 k(225.0)240.0 k2000 k1200 k

600 k

400 k

300.0 k200.0 k3000 k(2880)
3600 k2400 k1500 k(1440)
1800 k

720 k

480 k

360.0 k240.0 k

900 k

600 k

450 k

300.0 k200.0 k(192.0)240.0 k3000 k4.80 M

4.00 M

2400 k1200 k

800 k

600 k

400 k

320.0 k6.00 M

5.00 M

3000 k1500 k1000 k

750 k

500 k

400 k

7.20 M

6.00 M

3600 k1800 k1200 k

900 k

600 k

480 k

9.60 M

8.00 M

4.80 M

2400 k1600 k1200 k

800 k

640 k

12.00 M

10.00 M

6.00 M

3000 k2000 k1500 k1000 k800 k15.00 M
(14.40)18.00 M

12.00 M

7.20 M

3600 k2400 k1800 k1200 k960 k2400 k(2250)
2400 k1500 k1200 k1600 k1400 k(1280)
1500 k(1440)
1600 k

20.00 M
(19.20)24.00 M
(21.60)24.00 M
30.00 M
(28.80)36.00 M

15.00 M

9.00 M

4.50 M

3000 k

16.00 M

9.60 M

4.80 M

3200 k18.00 M

12.00 M
(10.80)12.00 M

5.60 M(5.40)
6.00 M

3600 k
4.00 M

2800 k(2700)
3000 k

15.00 M
(14.40)18.00 M

7.20 M

4.80 M

3600 k2400 k

9.00 M

6.00 M

4.50 M

3000 k

9.60 M

6.40 M

4.80 M

3200 k12.00 M
(10.80)12.00 M

7.20 M

3600 k

20.00 M
24.00 M
30.00 M

1800 k2000 k2000 k(1920)
2400 k

40.0 M(38.4)
45.0 M(43.2)
48.0 M

32.00 M

40.0 M

20.00 M
(19.20)24.00 M
(21.60)24.00 M

8.00 M

5.60 M(5.40)
6.00 M

4.00 M

2800 k(2560)
3000 k(2880)
3200 k

60.0 M

50.0 M

30.00 M

15.00 M

10.00 M

7.50 M

5.00 M

4.00 M

72.0 M

60.0 M

36.00 M

18.00 M

12.00 M

9.00 M

6.00 M

4.80 M

96.0 M

80.0 M

48.0 M

24.00 M

16.00 M

12.00 M

8.00 M

6.40 M

120.0 M

100.0 M

60.0 M

30.00 M

20.00 M

15.00 M

10.00 M

8.00 M

150.0 M
(144.0)180.0 M

120.0 M

72.0 M

36.00 M

24.00 M

18.00 M

12.00 M

9.60 M

150.0 M

90.0 M

45.0 M

30.00 M

15.00 M

12.00 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

160.0 M

96.0 M

48.0 M

32.00 M

24.00 M
(22.50)24.00 M

16.00 M

180.0 M

120.0 M(108.0)120.0 M

56.0 M(54.0)
60.0 M

36.00 M

18.00 M

40.0 M

28.00 M
(27.00)30.00 M

14.00 M(12.80)15.00 M
(14.40)16.00 M

300.0 M
(288.0)360.0 M

240.0 M

72.0 M

48.0 M

36.00 M

24.00 M

300.0 M

150.0 M
(144.0)180.0 M

90.0 M

60.0 M

45.0 M

30.00 M

20.00 M
(19.20)24.00 M

480 M

400 M

240.0 M

120.0 M

80.0 M

60.0 M

40.0 M

32.00 M

720 M

600 M

360.0 M

180.0 M

120.0 M

90.0 M

60.0 M

48.0 M

36.00 M

200.0 M

<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW

70

20.00 Mrate
[W]
4.50 k
(4.36)
5.60 k(5.24)
7.20 k(6.55)
7.20 k
(6.98)
9.00 k(8.73)
12.00 k(10.47)×0.1
14.00 k(13.09)18.00 k(17.45)24.00 k(21.82)28.00 k(26.18)36.00 k(34.91)45.0 k
(43.6)
56.0 k
(52.4)
72.0 k
(65.5)
72.0 k(69.8)
90.0 k(87.3)
120.0 k(104.7)×1
140.0 k(130.9)180.0 k(174.5)240.0 k(218.2)280.0 k(261.8)360.0 k(349.1)
450 k
(436)
560 k
(524)
720 k
(655)
720 k
(698)
800 k
(785)
900 k
(873)
1200 k(1047)
×10
1400 k(1309)
1400 k(1396)
1600 k(1571)
1800 k(1745)
2400 k(2182)
2800 k(2618)
3600 k(3491)
4.50 M(4.36)
5.60 M(5.24)
6.40 M(6.55)
7.20 M(6.98)
8.00 M(7.85)
8.40 M(8.73)
10.00 M
(10.47)×100
14.00 M
(13.09)18.00 M
(17.45)28.00 M
(26.18)×1000<Note 2>
In case the voltage range and the current range
were set asof an upper table and an
output pulse unit (a four-step setup ispossible) is set as the fastest. The output
pulse width of a pulse output is set to 100 to
130ms. (Usually 240 to 260ms)

SQLC-215-133
Appendix table 1-3
Active power range, watt-hour multiplier rate table (3-phase 3-wire)
V range
600V
600V
500V
300V
(VT460/110V) (VT440/110V) (VT380/110V) (VT220/110V)
Multiplierrate
A range
[W]
[W]
[W]
[W]
5A
6A
7.5A
8A
10A
×0.1

12A
15A
20A
25A
30A
40A
50A
60A
75A
80A
100A

×1

120A
150A
200A
250A
300A
400A
500A
600A
750A
800A
900A
1000A

×10

1200A
1500A
1600A
1800A
2000A
2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A×100

12000A
15000A
20000A×1000

30000A

4.20 k
(4.18)
5.60 k
(5.02)
6.40 k(6.27)
7.20 k
(6.69)
8.40 k
(8.36)
10.00 k(10.04)14.00 k(12.55)18.00 k(16.73)24.00 k(20.91)28.00 k(25.09)36.00 k(33.45)42.0 k(41.8)
56.0 k(50.2)
64.0 k
(62.7)
72.0 k(66.9)
84.0 k(83.6)
100.0 k(100.4)140.0 k(125.5)180.0 k(167.3)240.0 k(209.1)280.0 k(250.9)360.0 k(334.5)
420 k
(418)
560 k
(502)
640 k
(627)
720 k
(669)
800 k
(753)
840 k
(836)
1000 k(1004)
1400 k(1255)
1400 k(1338)
1600 k(1505)
1800 k(1673)
2400 k(2091)
2800 k(2509)
3600 k(3345)
4.20 M(4.18)
5.60 M(5.02)
6.40 M(6.27)
7.20 M(6.69)
8.00 M(7.53)
8.40 M(8.36)
10.00 M
(10.04)14.00 M
(12.55)18.00 M
(16.73)28.00 M
(25.09)4.00 k4.80 k
6.00 k
6.40 k
8.00 k
9.60 k12.00 k16.00 k20.00 k24.00 k32.00 k40.0 k
48.0 k
60.0 k
64.0 k
80.0 k
96.0 k
120.0 k160.0 k200.0 k240.0 k320.0 k
400 k
480 k
600 k
640 k
720 k
800 k
960 k
1200 k1400 k(1280)
1500 k(1440)
1600 k2000 k2400 k3200 k
4.00 M
4.80 M
6.00 M
6.40 M
7.20 M
8.00 M
9.60 M
12.00 M
16.00 M
24.00 M

3600 k(3455)
4.20 k
(4.15)
5.60 k
(5.18)
5.60 k
(5.53)
7.20 k
(6.91)
8.40 k
(8.29)
10.00 k(10.36)14.00 k(13.82)18.00 k(17.27)20.00 k(20.73)28.00 k(27.64)36.00 k(34.55)42.0 k(41.5)
56.0 k
(51.8)
56.0 k(55.3)
72.0 k(69.1)
84.0 k(82.9)
100.0 k(103.6)140.0 k(138.2)180.0 k(172.7)200.0 k(207.3)280.0 k(276.4)360.0 k(345.5)
420 k
(415)
560 k
(518)
560 k
(553)
640 k
(622)
720 k
(691)
840 k
(829)
1000 k(1036)
1200 k(1105)
1400 k(1244)
1400 k(1382)
1800 k(1727)
2000 k(2073)
2800 k(2764)
3600 k(3455)
4.20 M(4.15)
5.60 M(5.18)
5.60 M(5.53)
6.40 M(6.22)
7.20 M(6.91)
8.40 M(8.29)
10.00 M
(10.36)14.00 M
(13.82)20.00 M
(20.73)150V(110V)
[W]

2000

1000

2400

1200

3000

1500

3200

1600

4.00 k

2000

4.80 k

2400

6.00 k

3000

8.00 k

4.00 k

10.00 k5.00 k12.00 k6.00 kMultiplier
rate
×0.01

16.00 k8.00 k

20.00 k10.00 k24.00 k12.00 k30.00 k15.00 k32.00 k16.00 k40.0 k

20.00 k48.0 k

24.00 k60.0 k30.00 k80.0 k

40.0 k

100.0 k50.0 k

120.0 k60.0 k

×0.1

160.0 k80.0 k

200.0 k100.0 k240.0 k120.0 k300.0 k150.0 k320.0 k160.0 k360.0 k180.0 k400 k200.0 k480 k240.0 k600 k300.0 k640 k320.0 k720 k360.0 k

800 k

400 k

1000 k500 k1200 k600 k×11600 k800 k2000 k1000 k2400 k1200 k

3000 k1500 k3200 k1600 k3600 k1800 k

4.00 M

2000 k4.80 M

2400 k

6.00 M

3000 k

8.00 M

4.00 M

12.00 M

6.00 M×10

×100<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW71<Note 2>
In case the voltage range and the current range
were set asof an upper table and an
output pulse unit (a four-step setup ispossible) is set as the fastest. The output
pulse width of a pulse output is set to 100 to
130ms. (Usually 240 to 260ms)

SQLC-215-133

Head Office: 11-13 Hitotsuya 1-chome, Adachi-ku, Tokyo, 121-8639 Japan.
Phone：03-3885-2411 , Fax：03-3858-3966

Kyoto Office : 1-19 Ichinobe-Nishikawahara, Jyoyou-shi, Kyoto, 610-0114 Japan.
Phone：0774-55-1391 , Fax：0774-54-1353
August 26, 201572