TeslatronPT™
Integrated Cryofree®
superconducting
magnet system

Applications

Electrical Transport Measurements
High magnetic fields and low temperatures for Hall effect andquantum Hall effect measurements.

Low Dimensional Physics
Investigate the effect of low temperature and high field on low
dimensional materials; nanowires, nanotubes, quantum dots and
2D electron gases.

Spintronics
Spin processing and data storage studies using high magnetic
fields and low temperatures.

Electrical Transport

Low Dimensional

Measurements

Physics

Spintronics

TeslatronPT / p2

TeslatronPT Platform

Features / Benefits- Large sample space- Low temperature inserts available – down to 25 mK- Low-noise, ESD protected measurement probes- Integrated variable temperature insert with static exchange gas
around the sample, providing sample temperatures between 1.5
and 300 K while operating the magnet- Wide range of standard magnets with fields up to 14 T with
vector magnet geometries available- Fine filament Nb3Sn superconducting wire offers the minimum
field hysteresis via remnant field, and reduces flux jumping at
low fields- Intelligent control of cryogenic and magnetic environments withMercuryiTC temperature controller and MercuryiPS magnet
power supply

Insert features and options:- DC and RF wiring to the sample- < 300 mK with the HelioxVT ³He refrigerator- < 25 mK with the cryogen free KelvinoxJT dilution refrigerator
system- LCC sample carriers with universal interface, easily
demountable for quick sample exchange. Compatible with low
temperature inserts.

TeslatronPT / p4

Key Specifications
Temperature range< 1.5 K to 300 K

Standard sample probe
temperature stability

50 mK

System cooldown

60 hours from room temperature to 4 KStandard sample
probe cooldown(standard probe loaded into cold VTI)

Magnetic field

8, 12, 14 and 6-1-1 T

Variable temperature insert
sample space diameter

50 mm< 2h from room temperature to < 5 K

TeslatronPT / p6

Service Support
Our support to you
Because Oxford Instruments is unique in designing andmanufacturing the complete system, we offer unrivalled support andexpertise for your TeslatronPT system through our regional Customer
Support teams backed by unmatched factory expertise.

Related Products

For more productinformation pleasecontact yourregional office:U.K., Europe,Middle East, Africa
Tel: +44 (0) 1865 393200
Email: nanoscience@oxinst.com

HelioxVT
This sample-in-vacuum ³He
refrigerator achieves less than
300 mK for more than 40 hrs or
can provide 50 µW of coolingpower at 350 mK for over 6 hrs.
The HelioxVT uses a cold gas
environment with a 50 mm access,therefore no need for a 1 K pot on
the insert.

AmericasTel: +1 800 447 4717
Email: nanoscience@oxinst.com

China
Tel: +86 (0) 400 678 0609
Email: nanoscience@oxinst.com

KelvinoxJTJapanThe Kelvinox JT is a small,
dipstick-style, dilution refrigerator,
which features a Joule-Thomson
condensation stage. It means theKelvinoxJT can operate in any 4
K environment – wet or dry – as it
does not rely on a 1 K pot.

Tel: +81 (0) 3 4510 3526
Email: nanoscience.jp@oxinst.com®

MercuryiTC
Cryogenic programmableintelligent temperature controller.

MercuryiPS
Superconducting intelligent
magnet power supply.

Sample Protect
Protect sensitive samples fromelectrostatic discharge.

Nanonis Tramea
Fully integrated measurement
ready solution for quantum
transport.

Visit nanoscience.oxinst.com/dry-systems/products/teslatronpt
or email nanoscience@oxinst.com
Main service locations: UK, USA, Germany, China, Japan and India© Oxford Instruments Nanotechnology Tools Ltd, 2023.
All rights reserved.

India
Tel: +91 8181017017
Email: nanoscience@oxinst.com

Republic of Korea
Tel: +82 (2) 2047 6466
Email: nanoscience@oxinst.com

Asia & ROWEmail: nanoscience@oxinst.com