XLC-219-056

INSTRUCTION MANUALDC RECEIVING METER
XLC-110HARDWARE MODEL A【Without backlight】
XLC-110L
HARDWARE MODEL D【White backlight】

XLC-219-056

Thank you for purchasing DAIICHI ELECTRONICS product.
Please read this instruction manual carefully before using.

Safety precautions
 Environment conditions
Please be sure to use this product in a place that meets the following conditions.
In places that do not meet this condition, it may cause malfunction or failure and product life decline.
 Within the range of ambient temperature -10 to +55 ℃, humidity exceeding 85% RH.
 Place free of dust, corrosive gas, salt and oily smoke. (Corrosive gas：SO2 / H2S, etc.)
 Location that is not affected by vibration and shock.
 Location that is not affected by external noise.
 Altitude 1000m or less.
 Outdoor use conditions
 These products are not a dustproof, waterproof, and splash proof construction.
Please avoid the place with much dust. Moreover, please install in the place not exposed to rain or waterdrop.
 Please do not install in the place where sunlight hits directly.
Discoloration and degradation of a name plate, and deformation of the case by the surface temperature rise
may occur.
 Mounting and wiring
Please refer to this instruction manual for mounting and the wiring.
 Please refer to connection diagram for the wiring.
 Please avoid hot line work.
 Please use an electrical wire size suitable with the rated current.
 Please check the tightening of the screw.
 Please attach the terminal cover after work is completed
 Preparation
 When connecting this product directly to the main power supply, attach an appropriate fuse to the outside.
 This product must be set before use. Please set correctly after reading this instruction manual.
 About use
Be careful with the following cautions during use.
 Use the input within the rated range.
 Close the switch cover when the unit is not operated.
 This product has a function to hold the maximum and minimum values depending on the measurement element.
This value is guaranteed for power failure and is not cleared by a power reset.
However, if no input is applied when the power is turned on, the minimum value may be updated.
To keep the past minimum value when turning on the power, apply the input within 1 second after turningon the power.
 Be careful not to touch any terminal when power is applied to the unit.
 Don't disassemble or modify this unit without any previous permission of our company.
Modifications may cause troubles, a fire, or other accidents. For specifications change,
etc., please contact us.
 SettingThis product requires setting and confirmation such as unit display before use. In the case of use in the initial
setting, there is no need of setting and confirmation. If the setting is wrong, the measurement and output may
not work properly. Read the instruction manual before setting.
 About dew condensation
If the temperature and humidity of an installation change rapidly when a product is a non-energization, thewaterdrop by dew condensation may adhere to a display inner side. (The display filter and the LCD surface stickand the pattern of the shape of a circle or an ellipse occur.)This phenomenon does not cause any trouble. Disappears when control power is applied for 2 hours.

1

XLC-219-056

 Maintenance and inspection
 Inspection in energized state is dangerous.
 Check that the measured values, scale, units, etc. are displayed correctly.
 Check that there is no discoloration of the LCD or damage to the case. Also check that there are noloose wiring or mounting screws.
 No replacement in periodic inspection.
 Please wipe off lightly with the dry soft cloth. Please do not use the organic solvent, chemicals, cleaners,
etc., such as an alcohol, for cleaning.
The LCD screen may light up during wiping, which is caused by static electricity on the filter.
If left for a while, it will discharge naturally and return to its original state.
Also, when pressing the filter, the filter may come into contact with the liquid crystal display surface,causing a round or elliptical pattern. Do not press the filter strongly.
 Storage
Please store in a place that meets the following conditions.
 Within the range of ambient temperature -20 to +70 ℃, humidity exceeding 5 to 90% RH.
 Daily average temperature 40 ℃ or less.
 Location corresponding to the usage environment and use conditions.
 Aluminum electrolytic capacitors are used in products. Please energize the power supply within one
year after purchase.
 Countermeasures against troubles
If trouble occurs within the warranty period, DAIICHI ELECTRONICS will repairs this product.
 Disposal
Please dispose this product as industrial waste (non-combustible).
Mercury parts and a nickel-cadmium battery are not used for this product.
 Warranty period
The warranty period of the product is one year after the date of delivery.
 Warranty scope
In the state of the normal use of product-specification within the range according to this instruction manual,
the trouble within the warranty period performs exchange or repair gratuitously.
However, the shipping expenses and the packing cost in the case of shipping obtain as payment on a customer.
And, if it corresponds to the next, it does not warrant.
 If it breaks down when converted or repaired except our company.
 Failures caused by violation of various conditions concerning use, storage, etc. specified by the Company.
 If the cause of trouble is based on cause other than this product.
 Transportation, movement, damage by falling, and trouble.
 Other cases where the supplier is not responsible, such as a natural disaster or disaster.
Only a product simplex is applied with a warranty. We cannot bear the responsibility such as the damage andthe loss that occurred by the use of this product and trouble or the loss of profits.
And, a warranty is effective only in Japan.
 Changes to the contents of the instruction manual
Please note that the contents of this instruction manual are subject to change without notice due to productimprovements.

2

XLC-219-056

Content

1. Instructions on the product1.1 Application ･･･････････････････････････････････････････････････････････････

4

1.2 Features ･･････････････････････････････････････････････････････････････････

4

2. Names and functions of component parts ･･････････････････････････････････････

5

3. Preparation
3.1 Mounting ･･････････････････････････････････････････････････････････････････

6

3.2 Connections ･･･････････････････････････････････････････････････････････････

6

3.2.1 Connection diagram ･････････････････････････････････････････････････････

6

3.2.2 Cautions on connections ････････････････････････････････････････････････

7

4. Operation ･･･････････････････････････････････････････････････････････････････

7

5. Display ･････････････････････････････････････････････････････････････････････

8

6. Setting
6.1 Function table and default setting ････････････････････････････････････････ 10
6.2 Example of setting function ･･･････････････････････････････････････････････ 13
6.3 Setting table ･････････････････････････････････････････････････････････････ 15
6.4 Measurement display-mode in detail explanation ････････････････････････････ 18
6.5 Setting detail explanatory ････････････････････････････････････････････････ 22
6.5.1 Setting mode 1 ･････････････････････････････････････････････････････････ 22
6.5.2 Setting mode 2 ･････････････････････････････････････････････････････････ 28
6.6 About the scale of bar graph ･･････････････････････････････････････････････ 39
7. Specification
7.1 Specification code, Type ･･････････････････････････････････････････････････ 40
7.2 Specification ･････････････････････････････････････････････････････････････ 41
7.3 Performance ･･･････････････････････････････････････････････････････････････ 42
7.4 Option specifications ･････････････････････････････････････････････････････ 44
8. Trouble shooting ････････････････････････････････････････････････････････････ 44
Attached chart 1.

Bar graph scale division details.

3

XLC-219-056

1. Instructions on the product1.1 Application
In measurement of various physical quantities, although they are conversion and displayed on an electric
signal, this product is used for the purpose of the display.
Three physical quantities can be displayed simultaneously. And, three analog outputs proportional to thephysical quantity are made. The set monitor united with the system is made.

1.2 Features
 The simultaneous measurement display of dc-input 3 circuit is possible at one set.
 The withstand voltage between inputs is 2000V.
 This product has a scaling function that can set the full scale to any measurement value.
 18 types of units can be set arbitrarily.
 Bar graph display (1 measurement) can be checked like meter.
 Upper limit and lower limit flicker setting is possible (with setting index).
This index can also be used as a control index when the flickering function is turned off.
 The maximum and the minimum value of a measurement display can be held.
 Three analog output circuits can be taken out. (option)
 Power supply is AC85V to 253V or DC80 to 143V (for both AC and DC uses).

DC20 to 56V can also make.

 The mounting method of this unit is compatible with the mounting method of conventional 110 squaremechanical meter. This unit is mounted at 2 diagonal points.
 With backlight function (XLC-110L)
Always-on, Auto off (after 5 minutes without operating), Always-off and brightness setting is possible.
LED: White4

XLC-219-056

2. Names and functions of component parts

Bar graph display
Indicates measuring values as
analog values. The measurement
factor displayed on bar graphwith a DISPLAY switch is changed.

Digital display section.
Three factors can be measured andmonitored concurrently.
(An INPUT display can set ON/OFF.)Sub-monitor(Left)
Main monitor

Sub-monitor(Right)Flicker setting index(In case flicker setting is OFF,it displays as a check index.)

Input 1 circuit：Only INPUT 1 is displayed.
Input 2 circuit：Only INPUT 1 and 2 is displayed.
Input 3 circuit：INPUT 1, 2, 3 is displayed.
Scale digit
This is set automatically byindication scaling setting.

－

Unit indication
The unit can be set up by unit
display setting.

This switch is used for confirming
the flicker value. If no operation
is done for 10 seconds, the flicker
value indication returns to thedisplay mode.
In setting mode, it uses it formoving down of set value.

DISPLAY
This switch is the changes the input
factor which carries out a bar graphdisplay.
Main monitor
Sub-monitor(Left)

SET

Sub-monitor(Right)

This switch is used to select thesetting mode. When turning on thisswitch continuously for longer than
3 seconds, the mode is switched fromthe display mode to the setting mode
1. This switch is also used for
deciding a setting value in thesetting mode.

If it continues pressing more than
3 seconds simultaneously with a
SET switch, it will become thesetting mode 2 from a display mode.
The function can be replaced with＋ switch by setting.
The setting mode returns to thedisplay mode after one action.

＋
This switch is checks the maximum value and the minimum
value of each input. Changes in order of the maximum value →minimum value. If no operation is done for 10 minutes, the maximum
value and minimum value indication returns to the display mode.
This switch is also used to moveing up a setting value in thesetting mode.
This function can be replaced with DISPLAY switch by setting.

5

RESET/SHIFTThis switch resets the maximum hold and the minimum
holding value. The maximum and the minimum holding
value are reset by continuation ON more than 1 second.
In setting mode, it is used for movement of an item.

XLC-219-056

3. Preparation
3.1 Mounting
Refer to the external dimensions and panel cut, and mount in the panel (with a thickness of 10 mm or less)with the M5 nut provided.
● Dimensions diagram
110

16.5

10.5

104.5

90±0.5190±0.5

1±
10
φ

φ99

110

7-M4screw

15

92-M5screw

5-M4screw

8-M3screw(Option)

2-φ7

Panel cut（Front view）● LCD viewing angle
Mounting: Since the contrast of the LCD changes depending on the viewing angle, mount it at the optimal angle.
(1) Wide viewing angle model(2) For upper case installation

75°
10°
60°

75°
75

(Side view)

75

°

°

60
°

(Top view)(Side view)● Mounting(1) Insert the product into the cut hole of thepanel from the front.

60°(Top view)
（2）

（1）

(2) Secure the product with the mounting M5 flangenut (accessory).
The tightening torque of the flange nut shouldbe 2.0 to 2.5 N･m.

3.2 Connections
3.2.1 Connection diagram
Input 1 circuit.
INPUT 1

Input 2 circuit.

＋－

INPUT 1
INPUT 2

AUX.SUPPLY＋ －～ ～RESET INPUT＋ －～ ～Input 3 circuit.
＋－＋
INPUT 2－＋
INPUT 3－
AUX.SUPPLY

＋
－
＋
－

INPUT 1

AUX.SUPPLY
＋
－

＋ －
～ ～

＋
－
＋
－

OUTPUT 1＋ －～ ～OUTPUT 1
OUTPUT 2

RESET INPUTRESET INPUT
＋ －
～ ～

＋ －
～ ～

Note(１) Output 1, output 2, output 3, reset input is an option.
6

＋
－
＋
－
＋
－

OUTPUT 1
OUTPUT 2
OUTPUT 3

XLC-219-056

3.2.2 Cautions on connections(1) Mount the terminal cover without fail for safety after the end of connections.
(2) Separate the input wiring and output wiring from each other without fail, and take a preventive measureagainst malfunction due to external noises.
(3) Connect the grounding terminal E (No. 5 terminal) to the ground without fail for enhancing the shieldeffect keep the grounding resistance between the grounding terminal and the ground to be lower than100Ω.
(4) Keep a distance as much as possible of between this unit and the circuit breaker as well as between thisunit and the relay contact signal line.
(5) No protection is necessary for this unit even if the transmission line may be affected by an induced
lightning surge or the like when transmitting an analog output to the receiver.
Mount an about 500V arrester or the like between the line surge protector and the ground as well as
between the transmission line and the ground on the receiver side for the purpose of protecting thedevices on the receiver side.

4. Operation
DISPLAYSET－＋RESET/SHIFT

Switches

Functions

SET

By pressing this switch for longer than 3 seconds, the mode is switched from display mode
to setting mode 1. By pressing this switch for longer than 3 seconds together withDISPLAY switch, the mode is switched from display mode to setting mode 2.
This mode can be reset to the display mode by DISPLAY switch.

－

Flicker value can be checked.
This mode can be reset to the display mode by DISPLAY switch.

＋

Maximum value and minimum value can be checked.
This function can be replaced with a DISPLAY switch.

RESET/SHIFTDISPLAY

By pressing this switch for longer than 1 seconds, reset of maximum and the minimum value canbe performed.
The input factor on which bar graph is displayed can be changed.
Its function can be replaced with ＋ switch.

● Convenient functions
Even if the unit is stopped operating, it returns to the display mode in 10 minutes.

7

XLC-219-056

5. Display(1) Example of display of input 1 circuit
Display
combination
Display scaling

Display

Pattern 1

Main monitor：Input 1(Set the INPUT display to ON.)

Input 1：0.0 to 100.0A
INPUT1
Measurement display
symbol of input 1.
During measurement,
it always indicates.
Indication off ispossible by setting.

Bar graph display[Main monitor]
The measurement value of input
1 is indicated by the analog.
Digital display[Main monitor]
The measurement value of
input 1 is displayed.

(2) Example of display of input 2 circuit
Display
combination
Display scaling

Pattern 2

Main monitor：Input 1 , Sub to monitor (Right)：Input 2(Set the INPUT display to auto off.)
Input 1：45.0 to 55.0Hz
Input 2：COS LEAD 0.500 to 1.000 to LAG 0.500① Bar graph display of the main monitor.
Bar graph display[Main monitor]
When it changes to"main monitor" with a
DISPLAY switch, themeasurement value of an
input 1 is indicated by
the analog.

Display

Digital display[Main monitor]
The measurement value of
input 1 is displayed.
Frequency display：3 digit(or 4 digit)

Press DISPLAY switch② Bar graph display of the sub-monitor (right).
Digital display[Sub-monitor (right)]
The measurement valueof input 2 is displayed.
Power factor display：
4 digit , position
fixation of a decimal
point.

Bar graph display[Sub-monitor (right)]
When it changes to"sub-monitor (right)" with a
DISPLAY switch, themeasurement value of an input
2 is indicated by the analog.
An underbar indicates.
The underbar of a sub-monitor(right) indicates.

LEAD or LAG

indicates.

It indicates, only if it changes to the input which is doing COS
display setting. It does not indicate, if it changes to other inputs.

8

XLC-219-056(3) Example of display of input 3 circuit
Display
combination
Display scaling

Pattern 1
Main monitor：Input 1 , Sub-monitor (Left)：Input 2 ,sub-monitor (Right)：Input 3(Set the INPUT display to ON.)
Input 1：0.0 to 100.0℃
Input 2：0.0 to 100.0℃
If display scaling of 3 inputs is the same
Input 3：0.0 to 100.0℃① Bar graph display of main monitor.
Bar graph display[Main monitor]
When it changes to "mainmonitor" with a DISPLAY
switch, the measurement valueof an input 1 is indicated by
the analog.

INPUT 1
Measurement display
symbol of input 1.
During measurement,
it always indicates.
Indication off ispossible by setting.

Digital display[Main monitor]
The measurement value of
input 1 is displayed.
Press DISPLAY switch② Bar graph display of the sub-monitor (left).

Display

Bar graph display[Sub-monitor (left)]
When it changes to"sub-monitor (left)" with a
DISPLAY switch, themeasurement value of an input
2 is indicated by the analog.
An underbar indicates.

INPUT 2
Measurement display
symbol of input 2.
During measurement,
it always indicates.
Indication off ispossible by setting.
The underbar of a
sub-monitor (left)
indicates.

Digital display[Sub-monitor (left)]
The measurement value of
input 2 is displayed.

Press DISPLAY switch③ Bar graph display of the sub-monitor (right).
Bar graph display[Sub-monitor (right)]
When it changes to"sub-monitor (right)" with a
DISPLAY switch, themeasurement value of an input
3 is indicated by the analog.
An underbar indicates.

INPUT 3
Measurement display
symbol of input 3.
During measurement,
it always indicates.
Indication off ispossible by setting.
The underbar of a
sub-monitor (right)
indicates.
Press DISPLAY switch

Digital display[Sub-monitor (right)]
The measurement value of
input 3 is displayed.

＊ This is a limitation to the display pattern 1, and in order to check which input it isindicating, it is indicating "INPUT1", "INPUT2", and "INPUT3".
However, Indication off is possible by setting.

9

XLC-219-056

6. Setting
6.1 Function table and default setting
At the case of shipment, it is a following default setting. Please set up according to an operating condition.
In addition, about setting products, it is shipped by the appointed set point.
Setting mode 1.

Function table.

Setting
No.

Function

111

Display combination
setting

112
113
114

Unit
displaysetting

115

INPUT display ON/OFF

121H
122L

Input 1Flickersetting

123

Input 2Flickersetting126

129

Upper limit
value
Lower limit
value
ON/OFF

127H
128L

Upper limit
value
Lower limit
value
ON/OFF

124H
125L

Input 1
Input 2
Input 3

Input 3Flickersetting

Upper limit
value
Lower limit
value
ON/OFF

Functional description

Default setting

Set the combination pattern of a digital
display. By patterns 1 to 6, it can change
Pattern 1
arbitrarily the position of the digital
display of a maximum of 3 circuits.
Set the unit display of an input 1.
With no unit display
Set the unit display of an input 2.
With no unit display
Set the unit display of an input 3.
With no unit display
Set the ON/OFF of an INPUT display.
However, only the pattern 1 of display
ONcombination is effective.
OFF except for pattern 1.
The upper limit flicker value of an input 1
100.0(100％ of display span)measurement display is set up.
The lower limit flicker value of an input 1
0.0
measurement display is set up.
(0％ of display span)
When input is more than detection or less thandetection setting value, this sets flicker
OFF
ON/OFF of digital display.
The upper limit flicker value of an input
100.0(100％ of display span)
2 measurement display is set up.
The lower limit flicker value of an input
0.0
2 measurement display is set up.
(0％ of display span)
When input is more than detection or less thandetection setting value, this sets flicker
OFF
ON/OFF of digital display.
The upper limit flicker value of an input
100.0(100％ of display span)
3 measurement display is set up.
The lower limit flicker value of an input
0.0
3 measurement display is set up.
(0％ of display span)
When input is more than detection or less thandetection setting value, this sets flicker
OFF
ON/OFF of digital display.

ImportantPage
setting
○

23
24

○
○
○

23
24

○

23
24

○
○

25
26

○
○
○

25
26

○
○
○

25
26

○

Set this function when changing the ＋
switch and DISPLAY switch.

131

DISPLAY switch
function change
setting

0

1

DISPLAY switch：Bar graph display
change＋ switch：Maximum and minimum value
display change

026

AUTO (Auto OFF)27

3（Middle）

27

DISPLAY switch：Maximum and minimum
value display change＋ switch：Bar graph display change

151
(２)
152
(２)

Backlight action

Set the backlight action from ON (always-on),
AUTO (auto off), and OFF (always-off).

Backlight brightness

Set the brightness of backlight.

Note(２) This can set only at the case of white backlight specification.

10

XLC-219-056

Setting mode 2.
Setting
No.

211b
212F
213P
214
215b
216F
217P
218
219b
21AF
21bP
21C

Function table.
Function

Input 1.
Displayscalingsetting
Input 2.
Displayscalingsetting
Input 3.
Displayscalingsetting

221b

BIAS
MAX.
Decimal point
COS ,Hz, var
BIAS
MAX.
Decimal point
COS ,Hz, var
BIAS
MAX.
Decimal point
COS ,Hz, var
BIAS

Input 1.
Input
calibration
222F

SPAN

223b

BIAS
Input 2.
Input
calibration

224F

SPAN

225b

BIAS
Input 3.
Input
calibration

226F

SPAN

231

Input 1232

Input
sensitivity

233
241A
(３)
242A
(３)
243A
(３)

Input 2
Input 3
Output 1

OutputfactorOutput 2
Output 3

Functional description

Default setting

Display bias value setting of an input 1
Display max. value setting of an input 1
Decimal point setting of an input 1
COS ，Hz，var display scaling setting of
input 1
Display bias value setting of an input 2
Display max. value setting of an input 2
Decimal point setting of an input 2
COS ，Hz，var display scaling setting of
input 2
Display bias value setting of an input 3
Display max. value setting of an input 3
Decimal point setting of an input 3
COS ，Hz，var display scaling setting of
input 3
Zero adjustment of the input 1 (INPUT1) at
the case of a bias input can be performed.
Display and output are adjusted
simultaneously.
The input 1 display and output span
adjustment at the case of an input apply canbe performed. It is effective if a display
wants to synchronize with other meter.
Zero adjustment of the input 2 (INPUT2) atthe case of a bias input can be performed.
Display and output are adjusted
simultaneously.
The input 2 display and output span
adjustment at the case of an input apply canbe performed. It is effective if a display
wants to synchronize with other meter.
Zero adjustment of the input 3 (INPUT3) atthe case of a bias input can be performed.
Display and output are adjusted
simultaneously.
The input 3 display and output span
adjustment at the case of an input apply canbe performed. It is effective if a display
wants to synchronize with other meter.
The full scale of input 1 bar graph display
is changed.
The full scale of input 2 bar graph display
is changed.
The full scale of input 3 bar graph display
is changed.
Can be select which input element to output
to output 1 (OUTPUT1).
Can be select which input element to output
to output 2 (OUTPUT2).
Can be select which input element to output
to output 3 (OUTPUT3).

0.0
100.0
□□□.□
0
(Standard scale)
0.0
100.0
□□□.□
0
(Standard scale)
0.0
100.0
□□□.□
0
(Standard scale)

○
○
○

29
30

○
○
○
○

29
30

○

0.00
31
0.00

0.00
31
0.00

0.00
31
0.00

100
32

100
100
Input 1 (INPUT1)○

Input 2 (INPUT2)○

Input 3 (INPUT3)○

Note(３) This can set only at the case of analog output (option) specification.

11

ImportantPage
setting
○
○
29
○
30
○

33

XLC-219-056

Setting
No.

Function

251b
(４)

BIAS

252F
(４)

Output 1
Outputcalibration

253b
(４)
254F
(４)

BIAS
Output 2
Outputcalibration

255b
(４)
256F
(４)

Output 3
Outputcalibration

Low input
cut

Displaydead band

Input 1
Input 2
Input 3
Input 1

Measurement
display
ON/OFF

283
291

Input 2
Input 3

281
282

SPAN

Input 1

263
271
272
273

SPAN

BIAS

261
262

SPAN

Input 2
Input 3Returns to default setting

Functional description

Default setting

Output 1 (OUTPUT1) zero adjustment at thecase of a bias input can be performed.
Span adjustment of an output 1 can be
performed at the case of an input apply.
Up to 99% can be set.
Sensitivity change of an output can also be
performed.
Output 2 (OUTPUT2) zero adjustment at thecase of a bias input can be performed.
Span adjustment of an output 2 can be
performed at the case of an input apply.
Up to 99% can be set.
Sensitivity change of an output can also be
performed.
Output 3 (OUTPUT3) zero adjustment at thecase of a bias input can be performed.
Span adjustment of an output 3 can be
performed at the case of an input apply.
Up to 99% can be set.
Sensitivity change of an output can also be
performed.
In case of 0 to N, -N to 0 to N (example：-100 to 0 to 100) of display scaling.
It does below 0.5％ of inputs to 0 display.
And, it makes analog output into a bias
value.
-N' to 0 to N (example：-10 to 0 to 100) and-N to 0 do not function.
In case the input is unsteady, this setting
can drop the sensitivity of a display.
ON/OFF of an input 1 measurement display is
set up.
ON/OFF of an input 2 measurement display is
set up.
ON/OFF of an input 3 measurement display is
set up.
Initializes the all settings.

４

Note( ) This can set only at the case of analog output (option) specification.

12

ImportantPage
setting

0.00
34
0.00

0.00
34
0.00

0.00
34
0.00

OFF
OFF

35

OFF
0.0
0.0
0.0

36

ON
ON

37

ON
－

38

XLC-219-056

6.2 Example of setting function
Functional
example ①

Settingfunction

Function
explanation

Input：4 to 20mA，Display：0.0 to 100.0A.
In below 4mA of inputs, a display indicates the value (example. input 0mA → display：-25.0A)of minus. If you want to fix indicated value to 0 at the case of the below 4mA of inputs.
Please use "low input cut setting" (setting No.261 to 263) in the setting mode 2.
→ Indicated value is fixed to 0 at the case of the minute electric input equivalent to less than
0.5％ of an input span.
(Example. In case of 4 to 20mA, less than 4.08mA → display：0.0A)
However, effective only at the case of display scaling 0 to N and -N to 0 to N (example：-100
to 0 to 100). For -N' to 0 to N (example：-10 to 0 to 100) and -N to 0, this setting does not
work.
Before ( Setting：OFF )After ( Setting：ON )DISPLAYDISPLAY

100.0A

100.0A

50.0A

50.0A

0.0A
0mA

4mA

12mA

20mA

INPUT

-25.0A

Settingfunction

4mA-25.0A

Input：4mA → Display：0.0A
Input：0mA → Display：-25.0A
Functional
example ②

0.0A
0mA

12mA

20mA

INPUT

Less than 4.08mA

Input：4mA → Display：0.0A
Input：0mA → Display：0.0A

Input：±60mV , Display：±100.0A
If you want to fix indicated value to 0 at the case of the minute electric input near 0mV of inputs.
Please use "low input cut setting" (setting No.261 to 263) in the setting mode 2.
→ Indicated value is fixed to 0 at the case of the minute electric input equivalent to less than
0.5％ of an input span.
(Example. In case of ±60mV, less than ±0.3mV → display：0.0A)
However, effective only at the case of display scaling 0 to N and -N to 0 to N (example：-100
to 0 to 100). For -N' to 0 to N (example：-10 to 0 to 100) and -N to 0, this setting does not
work.
Before ( Setting：OFF )After ( Setting：ON )Display

Display

100.0A

100.0A
Less than -0.3mV

Function
explanation

-60mV

-60mV
0

60mV

Input

60mV

Input

Less than 0.3mV-100.0A-100.0A

Input：0.3mV → Display：0.5A
Input：0mV→ Display：0.0A

Input：Less than 0.3mV → Display：0.0A
Input：0mV→ Display：0.0A

13

XLC-219-056

Functional
example ③
Settingfunction

Used to suppress the variation in the indicated value by input variation.
Please use "Display dead band setting" (setting No. 271 to 273) in the setting mode 2.
→ The sensitivity of a display is dropped and the variation in indicated value is suppressed.
The setting range becomes 0.0％ to 2.0％ (0.1％ step).
Before ( Setting：0.0％ )After ( Setting：1.0％ )Function
explanation

Example) Near 60.0A, it varies from 59.3A to 60.8A
Functional
example ④
Settingfunction

Example) Variation is suppressed of near 60.0A.

In case the span or zero has shifted at the indicated value.
Please use "input calibration setting" (setting No.221b to 226F) in the setting mode 2.
→ A span or zero is adjusted of indicated value. The setting range becomes -9.99％ to 9.99％ of
an input span (0.01％ step).
Before ( Setting：BIAS 0.00％)
After ( Setting：BIAS -0.50％)

Function
explanation

Example) At the case of 4mA of inputs, it isdisplayed as place 0.5A whose indicatedvalue is 0.0A.

14

Example) At the case of 4mA of inputs, indicatedvalue can be adjusted to 0.0A.

XLC-219-056

6.3 Setting table
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
<Caution> There is a setting item excepted by input circuits or measurement display ON/OFF setting.
(1) Setting of display combination and unit display.
Items

Set the display
combination.
(111)

Set the unit
display of input 1
(112)
(５)

Setting and operation procedures
Press SET for longer than 3 seconds(111)
Select the display combination pattern by ＋ and －Selected display combination pattern is entered

Page

Press SET

Press DISPLAY

23,24
Returns todisplay mode

Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Press SET
Selected unit is entered
Select the unit by ＋ and －Press DISPLAY

23,24

Returns to display mode

Set the unit
display of input 2
(113)
(５)

Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Press RESET/SHIFT
Select the unit by ＋ and －Press SET
(113)
Press DISPLAY
Returns to display mode
Selected unit is entered

23,24

Set the unit
display of input 3
(114)
(５)

Press SET for longer than 3 seconds
Press RESET/SHIFT
(111)
(112)
Press RESET/SHIFT
Press RESET/SHIFT
Select the unit by ＋ and －
(113)
(114)
Selected unit is entered
Press DISPLAY
Returns to display mode
Press SET

23,24

Set the INPUT
display ON/OFF
(115)
(６)

Press SET for longer than 3 seconds
Press RESET/SHIFT
Press RESET/SHIFT
(111)
(112)
Press RESET/SHIFT
Press RESET/SHIFT
(113)
(114)
(115)
Press SET
Select an INPUT display ON or OFF by ＋ and －Press DISPLAY

Selected ON/OFF setting is entered

Returns to display mode

Note(５) While doing COS and Hz and var display setting, there is a setting item excepted.
Note(６) Only the display combination pattern 1 is effective. Setting is excepted except pattern 1.

15

23,24

XLC-219-056(2) Flicker setting
Items

Setting of upper
limit flicker
value of input 1.
(121H)

Setting and operation procedures

Page

Press ＋ and RESET/SHIFT together
Press SET for longer than 3 seconds(111)(121H)
Select an upper limit flicker value by ＋ and －Press SET
25,26
Selected upper limit flicker value is entered

Press DISPLAY

Returns to display mode

Setting of lower
limit flicker
value of input 1.
(122L)

Press SET for longer than 3 seconds
Press ＋ and RESET/SHIFT together(111)(121H)
Select an lower limit flicker value by ＋ and －Press RESET/SHIFT(122L)
Press SET
Selected lower limit flicker value is entered
Press DISPLAY

25,26

Returns to display mode

Setting of input 1
flicker ON/OFF.
(123)

Press SET for longer than 3 seconds
Press ＋ and RESET/SHIFT together(111)(121H)
Press RESET/SHIFT
Press RESET/SHIFT(122L)(123)
Select an flicker ON or flicker OFF by ＋ and －Press SET
Selected flicker ON/OFF setting is entered

Press DISPLAY

25,26

Returns to display
mode(3) Functional exchange setting of DISPLAY and ＋ switch.
Items

Changes thefunction of
DISPLAY switch
and ＋ switch.
(131)

Setting and operation procedures

Page

Press SET for longer than 3 seconds
Press ＋ and RESET/SHIFT together(111)(121H)
Select an function by ＋ and －Press ＋ and RESET/SHIFT together(131)
Selected function is entered
Press DISPLAY
Press SET
Returns to display mode

26

(Explanation)
Display in setting 0： DISPLAY Bar graph display change.
＋ Maximum value, minimum value display change.
Display in setting 1： DISPLAY Maximum value, minimum value display change.
＋ Bar graph display change.

16

XLC-219-056(4) Backlight setting
Items

Setting and operation procedures

Page

Press ＋ and RESET/SHIFT together
Press SET for longer than 3 seconds(111)(121H)
Setting of
backlight action.
(151)

Press ＋ and RESET/SHIFT together
Press ＋ and RESET/SHIFT together
(131)
(151)
Select an backlight action by ＋ and －Press SET
Selected backlight action is entered

Setting of
backlight
brightness.
(152)

Press DISPLAY

27

Returns to display mode

Press SET for longer than 3 seconds
Press ＋ and RESET/SHIFT together(111)(121H)
Press ＋ and RESET/SHIFT together
Press ＋ and RESET/SHIFT together
(131)
(151)
Press RESET/SHIFT
Select an backlight brightness by ＋ and －Press SET
(152)
Selected backlight brightness is entered
Press DISPLAY
Returns to display mode

27(5) Display scaling setting
Items
Setting of input 1
bias value.
(211b)(７)

Setting and operation procedures

Page

Press SET and DISPLAY together for longer than 3 seconds(211b)
Setting an bias value by ＋ and －Press SET
Setting bias value is entered

29,30

Press DISPLAY

Setting of input 1
max. value.
(212F)(７)

Setting of input 1
decimal point
position.
(213P)(７)

Returns to display mode

Press SET and DISPLAY together for longer than 3 seconds(211b)
Setting an max. value by ＋ and －Press SET
Press RESET/SHIFT(212F)
Setting max. value is entered
Press DISPLAY
Returns to display mode
Press RESET/SHIFT
Press SET and DISPLAY together for longer than 3 seconds(211b)
Press RESET/SHIFT
Setting an decimal point position by ＋ and －(212F)(213P)
Press SET
Setting decimal point position is entered
Press DISPLAY

29,30

29,30

Returns to display mode

Setting of input 1
COS , Hz, var.
(214)

Press SET and DISPLAY together for longer than 3 seconds
Press RESET/SHIFT(211b)
Press RESET/SHIFT
Press RESET/SHIFT(212F)(213P)(214)
Select an COS ，Hz, var by ＋ and －Press SET
Selected COS ，Hz, var is entered

Press DISPLAY

29,30

Returns to display mode

Note(７) The bias value and max. value and decimal-point position setting (211b to 213P) of an input 1 is
excepted, at the case of COS , Hz, var display (214) setting of input 1.

17

XLC-219-056

6.4 Measurement display-mode in detail explanation■ If a power supply is put into this product, it will become a measurement display mode.
Digital-display section[Main monitor, Sub-monitor (left), Sub-monitor (right)]
The measurement display of three factors can be performed simultaneously.
〔If display position is changed. To the setting mode 1.〕

・Iput 2 (INPUT2), input 3(INPUT3), are different by
the number of input circuits.
Sub-monitor (Right)
The measurement display of an
input 3 (INPUT3).
〔If display scaling is changed.
To the setting mode 2.〕

Sub-monitor (Left)
The measurement display of an
input 2 (INPUT2).
〔If display scaling is changed.
To the setting mode 2.〕

Main monitor
The measurement display of an
input 1 (INPUT1).
〔If display scaling is changed.
To the setting mode 2.〕

Bar graph display
A display factor is changed witha DISPLAY switch.
Flicker setting index (L)In case flicker ON/OFF setting
is OFF, it displays as a index.
〔If flicker setting is changed.
To the setting mode 1.〕

Numerical scale
Set the automatically by display
scaling.
〔If scale (sensitivity) of bar
graph is changed. To the setting
mode 2.〕Flicker setting index (H)■ Switch operation from a measurement display mode.
－

Flicker value display

DISPLAY＋
Displaymode

Maximum value, minimum value display (８)DISPLAYRESET/SHIFT
1 second

Maximum value, minimum value reset

DISPLAY

Bar graph display switching of an input factor (８)Note(８) If it is made the next setting by " DISPLAY switch-function exchange setting" and DISPLAY ispressed, the maximum value and the minimum value will indicate.
＋ ：Bar graph display change. DISPLAY ：Maximum value, minimum value display change.
・By continuing pressing a SET switch for 3 seconds, it becomes the setting mode 1. ⇒ 22 pages
・Press and hold the SET switch and DISPLAY switch at the same time for 3 seconds or more to entersetting mode 2. ⇒ 28 pages<Caution> There is a display item excepted by the number of input circuits or measurement display ON/OFF setting.
・In case of input 1 circuit specification, there is no display item of input 2 and input 3.
・In case of input 2 circuit specification, there is no display item of input 3.
・Please keep in mind that not all measurement values display in case a measurement display is OFF
altogether.

18

XLC-219-056

Operation from a measurement display mode
(1) Flicker value display
The check of the flicker value of an input factor can be performed.
Upper limit
flicker valueof input 1－

Display

Displaymode－

Lower limit
flicker valueof input 1－

Upper limit
flicker valueof input 3－

Upper limit
flicker valueof input 2－

Lower limit
flicker valueof input 2－－
DISPLAYor
Non-operationof 10 seconds

Lower limit
flicker valueof input 3

Operation

Whenever press the － switch from a display mode, it indicates the upper limit and lower
limit flicker value of an input factor.

Reset method

Press the DISPLAY
10 seconds.

switch or returns to a measurement display mode by no operating it for･The upper limit flicker-value
display of an input 1.

･The lower limit flicker-value
display of an input 1.

･The display in the state where
the flicker was excepted.

Upper limit.

Setting index.

Lower limit.

Setting index is not displayed.

Lower limit flicker value.

OFF：When a flicker value is
excepted.

Setting index.

Display
screenInput factor.
Upper limit flicker value.

19

XLC-219-056(2) Maximum value, minimum value display
The check of the maximum value and minimum value of an input factor can be performed.
The value is held even if maximum value and the minimum value turn off a power supply.

(９ )Maximum valueof input 1＋

Display

Operation

Display
mode

(９ )
＋

Minimum valueof input 1(９ )＋Maximum valueof input 2＋ (９ )DISPLAYor
Non-operationof 10 minutes(９ )＋Minimum valueof input 3(９ )＋Maximum valueof input 3(９ )＋Minimum valueof input 2

Whenever it presses ＋ switch from a display mode, the maximum and the minimum value of an input
factor are displayed.
It judges value with a value including polarity, for the maximum value and minimum value.
Example) In -10.0A and 5.0A, the maximum value is set to 5.0A and the minimum value is set
to -10.0A.
Note(９) If it is made the next setting by " DISPLAY switch-function exchange setting" andDISPLAY is pressed, the maximum value and the minimum value will indicate.
＋ ：Bar graph display change. DISPLAY ：Maximum value, minimum value display change.

Reset method

Presses the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
･Maximum value display of input 1･Minimum value display of input 1

Maximum value display [最大]Maximum value

Minimum value display [最小]Minimum value

The maximum and theminimum are displayedin Japanese.

Display
screen

最大 ：Maximum
最小 ：Minimum

20

XLC-219-056(3) Maximum and minimum value reset.
The maximum and the minimum holding value of an input factor are reset.
Resetprocess
Reset method
After reset

Displaymode

RESET/SHIFT
Pushes more than
1 second.

Maximum value and minimum value
reset of all input factors

During a display mode or maximum and minimum value display, a RESET/SHIFT switch is pressed forthe more than 1 seconds. The maximum and the minimum holding value of all input factors are reset.
Maximum value and minimum value are in agreement with the present input value after reset.
・Maximum value and minimum value reset display
A RESET/SHIFT switch is pressed more than 1 second.
Flashing the 最大 (maximum) and 最小 (minimum) for 3
seconds simultaneously.

Screendisplay(Pattern 1)

The maximum and the minimum are displayed in Japanese.
最大 ：Maximum
最小 ：Minimum(4) Bar graph display change
The input factor that carries out a bar graph display can be changed.
(10)Displaychangeprocess

DisplaychangemethodDISPLAY
10

Bar graph displayof main monitor( )DISPLAY

Bar graph displayof sub-monitor(left)(10)
DISPLAY

Bar graph displayof sub-monitor(right)Whenever press the DISPLAY switch in a display mode, the bar graphical representation of an input
factor changes.
Note(10) If it is made the next setting by " DISPLAY switch-function exchange setting" and＋ is pressed, the bar graphical representation of an input factor changes.
＋ ：Bar graph display change. DISPLAY ：Maximum value, minimum value display change.
・Bar graph display of main
monitor・Bar graph display of
sub-monitor (left)
Underbar lighting of a
sub-monitor (left)

Screendisplay(Pattern 1)

21

・Bar graph display of
sub-monitor (right)
Underbar lighting of a
sub-monitor (right)

XLC-219-056

6.5 Setting detail explanatory
6.5.1 Setting mode 1SET
3 seconds
DISPLAYRESET/SHIFT and ＋

Setting No. 111 to 115
Display combination, unit display setting
RESET/SHIFT and ＋

DISPLAY
Measurement
displaymode

RESET/SHIFT and －Setting No. 131
DISPLAY switch function change setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 121H to 129
Flicker setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －RESET/SHIFT and －Setting No. 151 to 152 (11)Backlight setting

If a SET switch is pressed for by continuation more than 3 seconds, it will become the setting mode 1
from a display mode.
It moves a setting item with RESET/SHIFT and ＋ switch (or RESET/SHIFT and － switch).
If a DISPLAY switch is pressed, it will return to a display mode.
If ＋ and － switches are pressed together for longer than 3 seconds during setting, the present set
values only are reset to the default settings.
<Caution> There is a display item excepted by the number of input circuits or measurement display ON/OFF
setting.
・In case of input 1 circuit specification, there is no display item of input 2 and input 3.
・In case of input 2 circuit specification, there is no display item of input 3.
・Please keep in mind that not all measurement values display in case a measurement display is
OFF altogether.
Note(11) Only white backlight specifications can be set.

22

XLC-219-056(1) Display combination，Unit display setting (Setting No. 111 to 115)
It can perform setting of a unit display of display combination and an input factor.
And, it can perform setting which erases a display of INPUT.
RESET/SHIFT111
Display
combination

RESET/SHIFT

1：Pattern 1＋

Setting
process

112
Unit display of
input 1RESET/SHIFT

0：No display

－

＋

＋

－

113
Unit display of
input 2RESET/SHIFT

0：No display
－

＋

＋

－

114
Unit display of
input 3RESET/SHIFT

115
INPUT display
ON/OFF

0：No display
－

＋

＋

－

ON
－

＋

＋

2：Pattern 2

＋

＋

－

－

－

OFF

－

The setting range of a unit display changes with display combination (111).
Please refer to a unit display list for details.

Pattern 1 only

5：Pattern 5
Default setting＋－

6：Pattern 6

Please refer to a display
combination list for details.

Setting No.
111
112
Setting item

113
114
115

Contents of setting
Display combination setting
Pattern 1,2
Unit display settingof an input 1.
Pattern 3 to 6
Pattern 3,4
Unit display settingof an input 2.
Pattern 1,2,5,6Pattern 5,6
Unit display settingof an input 3.
Pattern 1 to 4
INPUT display ON/OFF setting

Setting value possible range
Pattern 1 to 6
0 to 17 (Main monitor)
0 to 12 (Sub-monitor)
0 to 17 (Main monitor)
0 to 12 (Sub-monitor)
0 to 17 (Main monitor)
0 to 12 (Sub-monitor)
ON/OFF

Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
For shifting to the item of display combination setting and unit display setting,press ＋ and RESET/SHIFT or － and RESET/SHIFT together.
Setting display

Setting
methodSetting valuechangeUpdate ofsetting value
Return todefault setting
Reset method

Whenever it presses RESET/SHIFT switch, setting item changes.
Only the pattern 1 of INPUT display ON/OFF setting is effective.
Setting is excepted except pattern 1.
By display scaling setting in the setting mode 2, unit display setting (112
to 114) of an input factor which is doing COS (and Hz, and var) display
setting is excepted.
If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.

23

XLC-219-056

・Display combination setting
In the case of a pattern 1.
Setting No.

DisplayIn the case of a pattern 5.

Current setting

Indicates an input 2 ona sub-monitor (left).

Setting No.

Indicates aninput 3 on a
sub-monitor(right).

Indicates aninput 2 on a
sub-monitor(right).

Indicates aninput 1 on themain monitor.

Indicates aninput 3 on themain monitor.

New setting

・Unit display setting of input 1 (Pattern 1)
Setting No.

New setting

Current setting

Current setting

Indicates an input 1 ona sub-monitor (left).

New setting

・INPUT display ON/OFF setting (Pattern 1)
Setting No.

New setting

Current setting

Unit display

Display-position change is attained with display combination.
Pattern 1：Standard product ，Pattern 2 to 6：At the case of display-position designating.
1 input specification
2 input specification
3 input specification
INPUT
Sub
Sub
Sub
Sub
Sub
Sub
Pattern №
Main
Main
Main
monitor monitor
monitor monitor
monitor monitor display
Display
monitormonitormonitor(left) (right)(left) (right)(left) (right)
combination
Pattern 1 INPUT 1－－
INPUT 1 INPUT 2－INPUT 1 INPUT 2 INPUT 3 ON/OFF
list
Pattern 2
INPUT 1－
INPUT 2 INPUT 1 INPUT 3 INPUT 2
OFFPattern 3
INPUT 2 INPUT 1－
INPUT 2 INPUT 1 INPUT 3
OFFPattern 4
INPUT 2－
INPUT 1 INPUT 2 INPUT 3 INPUT 1
OFFPattern 5－
INPUT 1 INPUT 2 INPUT 3 INPUT 1 INPUT 2
OFFPattern 6－
INPUT 2 INPUT 1 INPUT 3 INPUT 2 INPUT 1
OFFWith display combination, the setting range of a unit display of an input factor changes.
INPUT 1
INPUT 2
INPUT 3Unitdisplay № Pattern 1,2 Pattern 3 to 6 Pattern 3,4 Pattern 1,2,5,6 Pattern 5,6 Pattern 1 to 40
Not display
Not display
Not display
Not display
Not display
Not display
1
A
A
A
A
A
A
2
kA
kA
kA
kA
kA
kA
3
V
V
V
V
V
V
4
kV
kV
kV
kV
kV
kV
5
W
℃
W
℃
W
℃
6
kW
％
kW
％
kW
％
Unit display
7
MW
m
MW
m
MW
m
list
8
℃
m3
℃
m3
℃
m3
9
％
m3/h
％
m3/h
％
m3/h
3
3
10
m
m /minmm /minm
m3/min
3
3
3
11
m
r/min
m
r/min
m
r/min
12
m3/h
min
m3/h
min
m3/h
min
13
m3/min－m3/min－m3/min
－
14
m/h
－
m/h
－
m/h
－
15
m/min
－
m/min
－
m/min
－
16
r/min
－
r/min
－
r/min
－
17
min
－
min
－
min
－
24

XLC-219-056(2) Flicker setting (Setting № 121H to 129)
The upper limit and low limit setting of the flicker of an input factor, and ON/OFF setting can be performed.
However, setting of an upper limit value and a lower limit value turns into setting in the range of an upper
limit value ＞ lower limit value.
RESET/SHIFT
121H
Upper limit
flicker valueof input 1

RESET/SHIFT

122L
Lower limit
flicker valueof input 1

124H
127H similitude

RESET/
SHIFT

123

124H to 126

127H to 129

Flicker ON/OFFof input 1

Setting itemof input 2

Setting itemof input 3

・124H
・125L
・126

・127H
・128L
・129

(121H to 123 similitude)(121H to 123 similitude)126
129 similitude

125L
128L similitude-20％ of span

OFF
OFF

Setting
process

＋

－

＋

－

－

-20％ of span
ON

－

＋

－

100％ of span

0％ of span

＋

＋

－

100％ of span

＋

＋

－

Setting No.
121H
122L
123
124H
125L
126
127H
128L
129

Setting display

Setting valuechangeDefault setting－

120％ of spanOFF

Setting
method－＋0％ of span＋

Setting item＋－

120％ of spanContents of a setting
Upper limit flicker setting of input 1
Lower limit flicker setting of input 1
Flicker ON/OFF setting of input 1
Upper limit flicker setting of input 2
Lower limit flicker setting of input 2
Flicker ON/OFF setting of input 2
Upper limit flicker setting of input 3
Lower limit flicker setting of input 3
Flicker ON/OFF setting of input 3

Setting value possible range-20％ to 120％ of display span. OFF (12)-20％ to 120％ of display span. OFF (12)ON/OFF-20％ to 120％ of display span. OFF (12)-20％ to 120％ of display span. OFF (12)ON/OFF-20％ to 120％ of display span. OFF (12)-20％ to 120％ of display span. OFF (12)ON/OFF

Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
For shifting to the item of flicker setting, press ＋ and RESET/SHIFT or －and RESET/SHIFT together.
Whenever it presses RESET/SHIFT switch, setting item changes.
If a ＋ switch or － switch is pressed, the set value will change.
The set point can change in high-speed operation gradually by continuingpressing ＋ or － switch.
A detecting function will be excepted if it is made setting which exceeds
120％ by upper limit flicker setting. (OFF，▲ disappears.)
A detecting function will be excepted if it is made setting which exceeds-20％ by lower limit flicker setting. (OFF，▲ disappears.)Note(12) Set value possible range when doing COS
100％ and OFF of display span.

Reset method

display setting, 0％ toUpdate ofsetting value

If a SET switch is pressed, the set point will update.

Return todefault setting

If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.
However, both an upper limit flicker value and a lower limit flicker value return
to a default setting.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.

25

XLC-219-056･Upper limit flicker value
setting of input 1.

･Lower limit flicker value
setting of input 1.

･Flicker ON/OFF setting of
input 1.

Setting №

Current setting

Setting №

Setting №

New setting

Setting index

Current setting

Current setting

Display

Setting index

New setting

New setting(3) DISPLAY switch function change setting (Setting No. № 131)
The function of DISPLAY switch and ＋ switch can be replaced.
Setting No. 131＋

Setting
process0
DISPLAY：Bar graph display
change＋：Maximum value and minimum
value change＋－

1
DISPLAY：Maximum value andminimum value change＋：Bar graph display change

Default setting－

Setting item

Setting No.

Contents of setting

Setting value possible range

131

Function exchange of DISPLAY switch and＋ switch

0，1

Setting display

Setting
methodSetting valuechangeUpdate ofsetting value
Return todefault setting

Reset method

Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
For shifting to the item of DISPLAY switch-function exchange setting, press＋ and RESET/SHIFT or － and RESET/SHIFT together.
If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・DISPLAY switch-function exchange setting
Setting No.

Current setting

Display
The setting are displayed in Japanese.
設定：Setting
New setting

26

XLC-219-056(4) Backlight setting (Setting No. 151 to 152)【White backlight only】
Set the action and brightness of backlight.

The backlight setting is only white backlight products.
◆ 151 Backlight action
The operation of the backlight can be selectedfrom ON (always on), AUTO (auto off), and OFF(always off).
If 5 minutes elapses without operating aswitch in case it is set as "AUTO (auto off)",
backlight is automatically off.
After that, backlight will be turned on ifeither of switches is operated.
Selection by ＋ and － , set value is updated
by SET .

RESET/SHIFT
151
Backlight
actionRESET/SHIFT

AUTO(Auto off)

152
Backlight
brightness

1

2
OFF
(Always off)

Setting
process◆ 152 Backlight brightness
The backlight brightness can be selected from5 levels from 1 to 5.
Backlight becomes the darkest if it is set as"1". Backlight becomes the brightest if it isset as "5".
Selection by ＋ and － , set value is updated
by SET .

3
ON(Always on)

4

5

Setting value
5
4
3
2
1

Setting item

Setting No.
151
152

Contents of setting
Backlight action
Backlight brightness

Brightness
BrightDark

Setting value possible range
AUTO（Auto OFF）
，OFF（always-off），ON（always-on）
1，2，3，4，5Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
Setting display

For shifting to the item of backlight setting, press ＋ and RESET/SHIFT or－ and RESET/SHIFT together.
Whenever it presses RESET/SHIFT switch, setting item changes.

Setting
methodSetting valuechangeUpdate ofsetting value
Return todefault setting

Reset method

If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
･ Backlight action setting
Setting No.
Current setting

Display
New setting

27

XLC-219-056

6.5.2 Setting mode 2
SET and DISPLAY
3 seconds
DISPLAYRESET/SHIFT and ＋

RESET/SHIFT and －Setting No. 211b to 21C
Display scaling setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 221b to 226F
Input calibration setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 231 to 233
Input sensitivity setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 241A to 243A
Output factor setting (13)RESET/SHIFT and ＋

Measurement
displaymode

DISPLAYRESET/SHIFT and －Setting No. 251b to 256F
Output calibration setting (13)RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 261 to 263
Low input cut setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 271 to 273
Display dead band setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 281 to 283
Measurement display ON/OFF setting
RESET/SHIFT and ＋

DISPLAYRESET/SHIFT and －Setting No. 291
Return to default setting.

If SET switch and DISPLAY switch are pressed together for longer than 3 seconds, it becomes the setting mode2 from measurement display mode.
It moves a setting item with RESET/SHIFT and ＋ switch (or RESET/SHIFT and － switch).
If a DISPLAY switch is pressed, it will return to a display mode.
If ＋ and － switches are pressed together for longer than 3 seconds during setting, the present set values
only are reset to the default settings.
<Caution> There is a display item excepted by the number of input circuits or measurement display ON/OFF setting.
・In case of input 1 circuit specification, there is no display item of input 2 and input 3.
・In case of input 2 circuit specification, there is no display item of input 3.
・In the case where a measurement display is OFF altogether, the next setting is possible.
Setting mode 2, "Returns to a default setting", "Measurement display ON/OFF setting".
Note(13) In case there is no analog output, there is no output setting.
28

XLC-219-056(1) Display scaling setting (Setting No. 211b to 21C)
Scaling setting of the indicated value of an input factor can be performed.
However, setting of an bias value and max. value turns into setting in the range of an bias value ＜ max. value.
RESET/SHIFT
211b
Bias value of
input 1RESET/SHIFT

212F
Max. value of
input 1215b
219b similitude

－

＋

－

＋

214
COS ,Hz,vardislay of input 1

RESET/
SHIFT

217b
21bP similitude

＋

－

0：OFF
＋

－

□□□.□

100.0
－

＋

＋

－

9999

1：Power factor
0.5 to 1 to 0.5

・215b
・216F
・217P
・218

・219b
・21AF
・21bP
・21C
211b to 214
similitude－Default setting

2：Power factor
0 to 1 to 0

－

□.□□□

219b to 21C
Setting itemof input 3

211b to 214
similitude

□□.□□
＋

Setting
process

－

－
＋

9998

215b to 218
Setting itemof input 2

218
21C similitude

□□□□

-9998

0.0
＋

213P
Decimal pointof input 1

216F
21AF similitude

-9999
＋

RESET/
SHIFT

＋

＋

－

－

3：Frequency
45 to 55Hz＋－

4：Frequency
55 to 65Hz＋－

5：Frequency
45 to 65Hz＋－

6：Reactive power
LEAD□ to 0 to LAG□

Setting No.
211b
212F
213P

Setting item

Contents of setting
Display bias value setting of input 1
Display max. value setting of input 1
Display decimal point setting of input 1

214

COS , Hz, var display setting of
input 1

215b
216F
217P

Display bias value setting of input 2
Display max. value setting of input 2
Display decimal point setting of input 2

218

COS , Hz, var display setting of
input 2

219b
21AF
21bP

Display bias value setting of input 3
Display max. value setting of input 3
Display decimal point setting of input 3

21C

COS , Hz, var display setting of
input 3

Setting value possible range-9999 to 9998 (var：LEAD 9999 to 1)-9998 to 9999 (var：LAG 1 to 9999)
No decimal point to 3 digits decimal point
COS ：0.5 to 1 to 0.5 ，0 to 1 to 0
Hz ：45 to 55Hz ，55 to 65Hz ，45 to 65Hz
Var ：LEAD□ to 0 to LAG□
(Set the □ in 211b to 213P)-9999 to 9998 (var：LEAD 9999 to 1)-9998 to 9999 (var：LAG 1 to 9999)
No decimal point to 3 digits decimal point
COS ：0.5 to 1 to 0.5 ，0 to 1 to 0
Hz ：45 to 55Hz ，55 to 65Hz ，45 to 65Hz
Var ：LEAD□ to 0 to LAG□
(Set the □ in 215b to 217P)-9999 to 9998 (var：LEAD 9999 to 1)-9998 to 9999 (var：LAG 1 to 9999)
No decimal point to 3 digits decimal point
COS ：0.5 to 1 to 0.5 ，0 to 1 to 0
Hz ：45 to 55Hz ，55 to 65Hz ，45 to 65Hz
Var ：LEAD□ to 0 to LAG□
(Set the □ in 219b to 21bP)

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of display scaling setting, press ＋ and RESET/SHIFT
Setting
methodSetting
displayor － and RESET/SHIFT together.
Whenever it presses RESET/SHIFT switch, setting item changes.
If COS or Hz display setting is being done by display scaling setting, thedisplay bias value of the input factor, the max. value, and decimal point
setting are excepted.
29

XLC-219-056

Setting
methodReset method

Setting valuechangeIf a ＋ switch or － switch is pressed, the set value will change. Set value
does high-speed action gradually by continuing pressing a ＋ or － switch.
If a 3 digits display and 4 digits display are switched at the case of Hz
display setting, if ＋ switch is pressed with a SET switch pressed, it willchange.
In a 4 digits display, it indicates it on the 4th digit of a digital displayas "4".

Update ofsetting value

If a SET switch is pressed, the set point will update. The upper limit (lower
limit) flicker value of the updated input factor, return to default setting.
(Flicker ON/OFF setting has no change.)

Return todefault
setting

If ＋ and － switches are pressed together for longer than 3 seconds during
setting, only as for the display scaling value of an input factor, and the upper
limit and the lower limit flicker value which have been set up now, both return
to a default setting.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・Bias setting of input 1

・Max. setting of input 1
Setting No.

Setting No.

Current setting

Current setting

New setting

New setting

・Decimal point setting of input 1
Setting No.
Current setting
New setting

Display

・COS setting of input 1
In case of COS ：0.5‐1‐0.5

・Hz setting of input 1
In case of frequency 45‐55Hz (3 digits display)

Setting No.

Setting No.

Current setting

Current setting

New setting

New setting

・var setting of input 1

In case of frequency 45‐55Hz (4 digits display)
Setting No.

Setting No.

Current setting

Current setting

New setting

4 digits display
New setting30

XLC-219-056(2) Input calibration setting (Setting No. 221b to 226F)
Indicated value adjustments (zero adjustment etc.) of an input factor can be performed.
RESET/SHIFT
221b
Bias value of
input 1RESET/SHIFT

222F
Span value of
input 1223b
225b similitude223b to 224F
Setting itemof input 2

224F
226F similitude-9.99％-9.99％

0.00％

0.00％

9.99％

9.99％

Setting item

Setting display

・225b
・226F

・223b
・224F

Setting
process

Setting No.
221b
222F
223b
224F
225b
226F

225b to 226F
Setting itemof input 3

Default setting

Contents of setting
Calibration (bias) setting of input 1
Calibration (span) setting of input 1
Calibration (bias) setting of input 2
Calibration (span) setting of input 2
Calibration (bias) setting of input 3
Calibration (span) setting of input 3

Setting value possible range-9.99％ to 9.99％ of input span
-9.99％ to 9.99％ of input span
-9.99％ to 9.99％ of input span
-9.99％ to 9.99％ of input span
-9.99％ to 9.99％ of input span
-9.99％ to 9.99％ of input span

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of input calibration setting, press ＋ and
RESET/SHIFT or － and RESET/SHIFT together.
Whenever it presses RESET/SHIFT switch, setting item changes.

Setting
methodReset method

Setting valuechangeIf a ＋ switch or － switch is pressed, the set value will change.
The set point can change in high-speed operation gradually by continuingpressing ＋ or － switch.

Update ofsetting value

If a SET switch is pressed, the set point will update.

Return todefault setting

If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
･Bias setting of input 1･Span setting of input 1

Setting No.

Setting No.

Setting value

Setting value

Display

Current measurement value

Current measurement value

31

XLC-219-056(3) Input sensitivity setting (Setting No. 231 to 233)
Sensitivity change of the bar graph display of an input factor can be performed.
RESET/SHIFT
231
Input sensitivity
setting of input 1

100％

RESET/SHIFT

232
Input sensitivity
setting of input 2

RESET/
SHIFT

233
Input sensitivity
setting of input 3

100％

100％

Setting
process
Default setting

Setting item

50％

50％

50％

1％

1％

1％

Setting No.
231
232
233

Contents of setting
Input sensitivity setting of input 1
Input sensitivity setting of input 2
Input sensitivity setting of input 3

Setting value possible range (14)1％ to 100％ of full scale
1％ to 100％ of full scale
1％ to 100％ of full scale

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of input sensitivity setting, press ＋ andSetting display

Setting
methodSetting valuechangeUpdate ofsetting value
Return todefault setting
Reset method

RESET/SHIFT or － and RESET/SHIFT together.
Whenever it presses RESET/SHIFT switch, setting item changes.
If COS or Hz display setting is being done by display scaling setting, only
the item of the input factor is excepted.
If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
･Input sensitivity setting of input 1
Setting No.

Display

Current setting value

New setting

Note(14) As for less than 39％ of input sensitivity setting, the accuracy of a bar graph display differs.

32

XLC-219-056(4) Output factor setting (Setting No. 241A to 243A)
The factor of the analog output to an input factor can be set up.
Analog output will not be outputted if it sets up without a factor.
(Example：It is set to 0mA at the case of 4 to 20mA.)
When there is no analog output, this setting item does not exist. And, setting of the input factor set as
OFF by measurement display ON/OFF is impossible.
RESET/SHIFT
241A
Input factor
setting of output 1

RESET/SHIFT

242A
Input factor
setting of output 2

1：Input 1

Setting
process

Setting itemRESET/SHIFT

1：Input 1

243A
Input factor
setting of output 3

1：Input 1
Default setting

2：Input 2

2：Input 2

2：Input 2

3：Input 3

3：Input 3

3：Input 3

0：No factor

0：No factor

0：No factor

Setting No.
241A
242A
243A

Setting display

Contents of setting
Input factor setting of output 1
Input factor setting of output 2
Input factor setting of output 3

Setting value possible range
Input 1 to 3，No factor
Input 1 to 3，No factor
Input 1 to 3，No factor

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of output factor setting, press ＋ and RESET/SHIFTor － and RESET/SHIFT together.
Whenever it presses RESET/SHIFT switch, setting item changes.

Setting
methodReset method

Setting valuechangeIf a ＋ switch or － switch is pressed, the set value will change.
Setting of the input factor set as OFF by measurement display ON/OFF is
impossible.

Update ofsetting value

If a SET switch is pressed, the set point will update.
Analog output will not be outputted if it sets up without a factor.
(Example：It is set to 0mA at the case of 4 to 20mA.)

Return todefault setting

If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.
However, it is set up without a factor when the input factor of a default
setting is OFF.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・Factor setting of an output 1
Setting No.
Current setting

Display
New setting

33

XLC-219-056(5) Output calibration setting (Setting No. 251b to 256F)
Adjustment of the output value of analog output can be performed.
In case there is no analog output, this setting item does not exist.
RESET/SHIFT
251b
Bias value of
output 1

RESET/SHIFT

252F
Span value of
output 1

253b
255b similitude-9.99％

0.00％

0.00％

9.99％

99.00％

Setting
process

Setting item

255b to 256F
Setting itemof output 3

254F
256F similitude-9.99％

Setting No.
251b
252F
253b
254F
255b
256F

253b to 254F
Setting itemof output 2

・255b
・256F

・253b
・254F

Default setting

Contents of setting
Calibration (bias) setting of output 1
Calibration (span) setting of output 1
Calibration (bias) setting of output 2
Calibration (span) setting of output 2
Calibration (bias) setting of output 3
Calibration (span) setting of output 3

Setting value possible range-9.99％ to 9.99％ for output span
-9.99％ to 99.00％ for output span (15)-9.99％ to 9.99％ for output span
-9.99％ to 99.00％ for output span (15)-9.99％ to 9.99％ for output span
-9.99％ to 99.00％ for output span (15)

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of output calibration setting, press ＋ andSetting display

Whenever it presses RESET/SHIFT switch, setting item changes.
By output factor setting, the setting item of the output set as "No factor"
is excepted.

Setting
methodReset method

RESET/SHIFT or － and RESET/SHIFT together.

Setting valuechangeIf a ＋ switch or － switch is pressed, the set value will change.
The set point can change in high-speed operation gradually by continuingpressing ＋ or － switch.

Update ofsetting value

If a SET switch is pressed, the set point will update.

Return todefault setting

If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
･Bias setting of output 1.

･Span setting of output 1.

Setting No.

Setting No.

Setting value

Setting value

Display

Current measurement value

Current measurement value

15

Note( ) As for more than 60.01％ of output calibration span setting, the accuracy of a analog output differs.
34

XLC-219-056(6) Low input cut setting (Setting No. 261 to 263)
Indicated value is fixed to 0 at the case of the minute input equivalent to less than 0.5％ of an input span.
However, effective only at the case of display scaling 0 to N, and -N to 0 to N (example：-100 to 0 to 100).
Don't function at the case of -N' to 0 to N (example：-10 to 0 to 100) and -N to 0.
RESET/SHIFT
261
Low input cut
setting of input 1

Setting
process

RESET/SHIFT

262
Low input cut
setting of input 2

OFF

RESET/
SHIFT

OFF

263
Low input cut
setting of input 3

OFFDefault settingON

Setting item

Setting No.
261
262
263

Setting display

ON

Contents of setting
Low input cut setting of input 1
Low input cut setting of input 2
Low input cut setting of input 3

ON

Setting value possible range
ON/OFF
ON/OFF
ON/OFF

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of low input cut setting, press ＋ and RESET/SHIFTor － and RESET/SHIFT together.

Setting
methodWhenever it presses RESET/SHIFT switch, setting item changes.
Setting valuechangeUpdate ofsetting value
Return todefault setting

Reset method

If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・Low input cut setting of input 1
Setting No.

Display

Current setting value

New setting

35

XLC-219-056(7) Display dead band setting (Setting No. 271 to 273)
Set to suppress the variation in the indicated value by input variation.
RESET/SHIFT
271
Display dead bandsetting of input 1

0.0％

RESET/
SHIFT

272
Display dead bandsetting of input 2

RESET/
SHIFT

0.0％

273
Display dead bandsetting of input 3

0.0％

Setting
process

Setting item

Default setting
1.0％

1.0％

1.0％

2.0％

2.0％

2.0％

Setting No.
271
272
273

Setting display

Contents of setting
Display dead band setting of input 1
Display dead band setting of input 2
Display dead band setting of input 3

Setting value possible range
0.0％ to 2.0％
0.0％ to 2.0％
0.0％ to 2.0％

If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of display dead band setting, press ＋ and
RESET/SHIFT or － and RESET/SHIFT together.

Setting
methodWhenever it presses RESET/SHIFT switch, setting item changes.
Setting valuechangeUpdate ofsetting value
Return todefault setting

Reset method

If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・Display dead band setting of input 1
Setting No.

Display

Current setting value

New setting36

XLC-219-056(8) Measurement display ON/OFF setting (Setting No. 281 to 283)
Set the measurement display ON/OFF of an input factor.
Related output elements cannot be set for input elements that are OFF by this setting.
And in case setting the input factor of OFF, the output is OFF.
If measurement display setting is ON from OFF, because an output is still 0FF, please redo setting of theoutput factor setting.
RESET/SHIFT
281
Measurement display
ON/OFF setting of
input 1

Setting
process

RESET/SHIFT

282
Measurement display
ON/OFF setting of
input 2RESET/SHIFT

283
Measurement display
ON/OFF setting of
input 3

ON

ON

ON

OFF

OFF

OFF
Default setting

Setting item

Setting No.
281
282
283

Setting display

Setting
methodIf SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of measurement display ON/OFF setting, press ＋ and
RESET/SHIFT or － and RESET/SHIFT together.

Setting valuechangeUpdate ofsetting value

If a ＋ switch or － switch is pressed, the set value will change.
If a SET switch is pressed, the set point will update.
If ＋ and － switches are pressed together for longer than 3 seconds during
setting, the present set values only are reset to the default setting values.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・Setting of measurement display ON/OFF of an input 1.
Setting No.

Display

Setting value possible range
ON/OFF
ON/OFF
ON/OFF

Whenever it presses RESET/SHIFT switch, setting item changes.

Return todefault setting
Reset method

Contents of setting
Measurement display ON/OFF setting of input 1
Measurement display ON/OFF setting of input 2
Measurement display ON/OFF setting of input 3

Current setting value

New setting37

XLC-219-056(9) Return to default setting (Setting No. 291)
Returns all settings to their default settings.
291
Default setting

Setting
process

Push SET for morethan 3 seconds.
Return to default setting

Setting item

Setting
methodSetting No.
291

Setting display

Return todefault setting
Reset method

Contents of setting
Return to default setting

Setting value possible range―
If SET switch and DISPLAY switch are pressed together for longer than 3
seconds, it becomes the setting mode 2 from measurement display mode.
For shifting to the item of "return to default setting", press ＋ and
RESET/SHIFT or － and RESET/SHIFT together.
SET switches are pressed for longer than 3 seconds, all the set points
return to an default setting.

Press the DISPLAY switch or returns to a measurement display mode by no operating it for
10 minutes.
・Return to default setting
Setting No.

Display

Lights when initialization is execute.

38

XLC-219-056

6.6 About the scale of bar graph(1) A bar graph scale turns into nearest scale including the full-scale value of a display.
It automatic-selects from the following standard scale, and displays.
And, a standard bar graph display becomes three kinds. (0 to N，-N to 0，-N to 0 to N)
Bar graph display of standard

At fragment swing
At both swing

Example) Fragment swing-9900≦N≦9900

Example) Both swing

In case of display scaling setting 0 to 2000W
of input 1.
Standard scale

0 to N or -N to 0
-N to 0 to N

In case of display scaling setting -100.0 to 100.0W
of input 1.

1 ，1.2 ，1.5 ，1.6 ，1.8 ，2 ，2.4 ，2.5 ，3 ，3.2 ，3.6 ，4 ，4.5 ，4.8 ，5 ，6 ，
6.4 ，7.2 ，7.5 ，8 ，9 ，9.6
Integral multiple of 10.(10ｎ)(2) In case a display full-scale value becomes between the above scale division by setting of a display
scaling. The nearest bar graph scale including the value is selected.
Example) In case of display scaling setting-98.0 to +98.0W of input 1.

Example) In case of display scaling setting
0 to 1900W of input 1.

The scale of 2000 is selected because there
is no scale 1900.

The scale of 100 is selected because there
is no scale 98.

(The nearest bar scale including 1900 values
is selected.)(The nearest bar scale including 98.0 values
is selected.)<Caution>・In case of the scale which does not contain 0, such as 10 to 2000, display as the standard bar graph scale is
0 to 2000.
・In case of unbalanced scales, such as -10 to 0 to 100, display as the standard bar graph scale is -100 to 0
to 100.
(3) Please refer to an attached chart 1 "Bar graph scale division details" about the details of each scale division.
39

XLC-219-056

7. Specification
7.1 Specification code, Type
①
Type

－

②

③

④

Hard
model

Input
circuit

Input
range

Product name
Type (Function)

Without backlightWith backlight①②

Hard model③

Input circuit④

Input range(Specification code for DC input)

⑤

⑥

⑦

⑧

⑤

⑥
External
Auxiliary
operation
supply
input

⑧

⑦
－

Analogoutput0

Mounting
position

DC receiving meter
XLC-110
XLC-110L
DC input × 3 circuit (MAX.)
Code
A
D
6
7
8
1
2
3
4
5
6
7
8
9
A

Model A , Without backlight
Model D , With backlight (White)
DC input × 1 circuit
DC input × 2 circuit
DC input × 3 circuit
DC1 to 5V
DC0 to 1V
DC0 to 5V
DC0 to 10V
DC4 to 20mA
DC0 to 1mA
DC0 to 5mA
DC0 to 10mA
DC0 to 16mA
DC0 to 20mA

Auxiliary supply

Z
1
2

Other (16)AC85 to 253V DC80 to 143V for both AC and DC uses
DC20 to 56V

External operation input

Z
0
2

Other
Nothing
External reset

Z
0
1
2
3
4
5

Other
Nothing
DC4 to 20mA
DC0 to 1mA
DC1 to 5V
DC0 to 5V
DC0 to 10V

Z
0F

Other (17)For upper installation (For lower view)
Wide viewing angle (Hard model D only)

Analog output

Mounting position (LCD view angle)

Note(16) 1) Input range standard range ･････ Voltage input：±50mV to ±300V
Current input：±500μA to ±50mA2）Input range special range ･･････ 1. In case all rating is not uniformity.
2. Current input：±100μA to ±499μA(The digital display accuracy changes to ±1.0％→±1.5％)
Note(17) 1) Analog output standard range ･･･ Voltage output：±100mV to ±10V
Current output：±500μA to +20mA，-10mA
The number of output circuits becomes the number of input
circuits, and uniformity.
2) Analog output special range ････ 1. In case all rating is not uniformity.
2. In case the number of inputs and the number of outputs are not
the same. (Example：3 input → 1 output)
3. Current output：±100μA to ±499μA(The analog output accuracy changes to ±0.5％→±1.0％)
40

XLC-219-056

7.2 Specification
Item
Number of input
circuits

Input range

Digital display
range

Bar graphdisplay range

Specification
Maximum of 3 circuits (Mutual insulates by AC2000V.)
Code

Input

1

DC1 to 5V

Input
resistance
About 1MΩ2

DC0 to 1V

About 1MΩ3

DC0 to 5V

About 1MΩ4

DC0 to 10V

About 1MΩ5

DC4 to 20mA

About 50Ω

6

DC0 to 1mA

About 1kΩ

7

DC0 to 5mA

About 200Ω

8

DC0 to 10mA

About 100Ω

9

DC0 to 16mA

About 50ΩA

DC0 to 20mA

About 50Ω

Z

Other

－

Remarks
Standard input range.

Voltage input ±50mV to ±300V
Current input ±500μA to ±50mASpecial input range.
1. In case rating is not the same.
2. Current input ±100μA to ±499μA(The digital display accuracy changes to ±1.0％ → ±1.5％)
Input serves as 3 circuit (maximum) same rating. (Standard)
Example) Input 1：4 to 20mA，Input 2：4 to 20mA，
Input 3：4 to 20mA
If the inputs are not the same, it will be "Z". (Special)
Example) Input 1：4 to 20mA，Input 2：0 to 1mA，
Input 3：1 to 5V
Arbitrarily setting of theposition of the number of digits
and decimal point.
4 digits fixation.
The position of decimal point isfixed.
3 digits or 4 digits fixation.
The position of decimal point isfixed.
Arbitrarily setting of theposition of the number of digits
and decimal point.

Digital display-9999 to 9999

Power factor(COS ) display(1) LEAD 0.500 to 1.000 to LAG 0.500(2) LEAD 0.000 to 1.000 to LAG 0.000

Frequency display(1) 45.0 to 55.0Hz or 45.00 to 55.00Hz(2) 55.0 to 65.0Hz or 55.00 to 65.00Hz(3) 45.0 to 65.0Hz or 45.00 to 65.00Hz

Reactive power
display(LEAD，LAG)

LEAD 9999 to 0 to LAG 9999

Maximum scalevalue

1 ，1.2 ，1.5 ，1.6 ，1.8 ，2 ，
2.4 ，2.5 ，3 ，3.2 ，3.6 ，4 ，
4.5 ，4.8 ，5 ，6 ，6.4 ，7.2 ，
7.5 ，8 ，9 ，9.6
Integral number times of 10.(10ｎ)However range of -9900≦N≦9900

Power factor(COS ) display(1) LEAD 0.5 to 1 to LAG 0.5(2) LEAD 0 to 1 to LAG 0

A scale value is fixed.
Only at the case of power factor
display selection, LEAD and LAG
displays.

Frequency display(1) 45 to 55Hz
(2) 55 to 65Hz
(3) 45 to 65Hz

A scale value is fixed.

Reactive power
display(LEAD，LAG)

LEAD□ to 0 to LAG□
□ is the same numerical value as theabove maximum scale.

However, the range of LEAD 9900 to
0 to LAG 9900.
Only at the case of reactive power
display selection, LEAD and LAG
indicates.

41

XLC-219-056

ItemStandard unit

LCD (18 types) (18)(1) A(2) kA(3) kV(4) kW(19)(5) m(6) m/h(19)(7) m/min (19)(8) min(9) m3(10) m3/h(11) m3/min(12) MW(19)(13) r/min(14) V(15) W(19)(16) ％(17) ℃(18) No unit

(1)
(2)
(3)
(4)
(5)
(6)
(7)
(8)
(9)
(10)
(11)
(12)
(13)
(14)
(15)
(16)
(17)
(18)

APm
bar
cm
COS
ELm
Hz
J
K
kg
kg/h
kg/m2
kg/m3
kL
kN
kPa
kvar
KW
L

Specification
Unit lettering display (55 type) (20)(19) L/h(37) Nm3/min(55)(20) L/min(38) N/m2(21) mA(39) N/mm2(22) mg/L(40) OPm(23) min-1(41) Pa(24) mL/min(42) pH(25) mm(43) ppm(26) m/h(21) (44) R(27) m/min (21) (45) rad(28) m/s(46) rpm(29) MV(47) SPm(30) m3/s(48) t(31) Mpa(49) t/h(32) Mvar(50) TPm(33) MW(21) (51) W(21)(34) N(52) YPm(21) (35) N･m(53) μm(36) Nm3/h(54) μS/cm

度

Note(18) LCD, Main monitor：18 types. Sub-monitor：13 types.
Note(19) LCD display is possible only for the main monitor. Sub-monitor cannot display of LCD.
Note(20) Lettering display. Main monitor：50 types, Sub-monitor：55 types.
Lettering character height. Main monitor：8.5mm, Sub-monitor：5mm.
The character color of lettering, Gray (DIC, The 13th edition, 541)
Note(21) A lettering display is possible only for a sub-monitor. The main monitor serves as a liquid crystal display.

7.3 Performance
Item
Accuracy of digital
display
Accuracy of bar graph
Influence of temperature

Conformity technicalstandard

Display updating time
Display device,
Display composition

Specification±1.0％±5.0％ (％ for span)
23±10℃ in accuracy
JIS C 1102-1：1997 ･･･ Direct acting indicating analogue electrical measuring
instruments and their accessories.
Part 1：Definitions and general requirements common to allparts.
JIS C 1102-2：1997 ･･･ Direct acting indicating analogue electrical measuring
instruments and their accessories.
Part 2：Special requirements for ammeters and voltmeters.
JIS C 1102-7：1997 ･･･ Direct acting indicating analogue electrical measuring
instruments and their accessories.
Part 7：Special requirements for multi-function instruments.
JIS C 1102-9：1997 ･･･ Direct acting indicating analogue electrical measuring
instruments and their accessories.
Part 9：Recommended test methods.
JIS C 1111：1989 ･････ Electrical measuring transducers for converting a.c.
electrical quantities into d.c. electrical quantities.
JIS C 1010-1：1998 ･･･ Safety requirements for electrical equipment formeasurement, control, and laboratory use.
Part 1：General requirements.
About 1 second (Bar graph of about 0.25 second)
Main monitor
Character height 10mm, 4 digits
Character height 6mm, 4 digits
LCD Sub-monitor (Left),(Right)
Bar graph
30 dots42

XLC-219-056

Item

Specification(1) AC85 to 253V 50/60Hz 10VA (Rated voltage AC100/110V, 200/220V)
XLC-110DC80 to 143V
5W (Rated voltage DC100/110V)(Without
for both AC and DC usesback light)(2) DC20 to 56V
6W (Rated voltage DC24/48V)(1) or (2).
(1) AC85 to 253V 50/60Hz 12VA (Rated voltage AC100/110V, 200/220V) Designate
XLC-110L
DC80 to 143V
6W (Rated voltage DC100/110V)
Auxiliary (Withfor both AC and DC usesback light)
supply(2) DC20 to 56V
7W (Rated voltage DC24/48V)
Rated voltage AC110V
Less than 5.2A (About 1.7ms)
Rated voltage AC220V
Less than 10.4A (About 1.7ms)
Rush current
Rated voltage DC110V
Less than 3.7A (About 1.7ms)(Time constant)
Rated voltage DC24V
Less than 5.5A (About 3.6ms)
Rated voltage DC48V
Less than 10.9A (About 3.6ms)
Voltage circuit
2 times 10 seconds, 1.2 times continuation of rated voltage
Current circuit
10 times 5 seconds, 1.2 times continuation of rated current
Overload capacity
1.5 times 10 seconds, 1.2 times continuation of rated voltage.
Auxiliary supply In case of DC110V, 1.5 times 10 seconds, 1.3 times continuationof rated voltage.
Between electric circuit and a case (ground).
Between input, output, auxiliary supply.
Above 50MΩ at DC500V megger.
Insulation resistance
Between inputs.
Between analog outputs.
Non-insulation (Minus common)
Between electric circuit and a case (ground).
Between input, output, auxiliary supply.
AC2000V (50/60Hz) 1 minuteWithstand voltage
Between inputs.
Between analog outputs.
Non-insulation (Minus common)
5kV 1.2/50μs Positive and negative
Impulse withstand voltage Between electric circuit and a case (ground).
polarity for each 3 times(1) Oscillatory surge voltage
If a vibration damping waveform (1 to 1.5MHz, peak voltage：2.5 to 3kV) is repeated
and added, error is less than ±10％.
Voltage circuit, current circuit (Common)
Auxiliary supply circuit (Normal / Common)(2) Square wave impulse noise
If a noise (1μs, 100ns width) is repeated and it adds for 5 minutes, error isless than ±10％.
Voltage, current circuit (Common)
Over 1.5kV
Auxiliary supply circuit (Normal / Common) Over 1.5kV
Noise-capacity
External operation input (Common)
Over 1.0kV
Analog output (Induction)
Over 1.0kV(3) Wave noise
If the intermittence irradiation of the wave (5W, 1m) of 150MHz, 400MHz bandis done, and if the intermittence irradiation of the wave (1m) of a cellular
phone is done, error is less than ±10％.
(4) Electrostatic noise
At the 8kV at power distribution, error is less than ±10％.
There needs to be no 10kV damage at the case of the non-power distribution.
Condenser charge form.
Vibration：Single amplitude 0.15mm, 10 to 55Hz, Each minute octave in 5 times sweep
Vibration, shock
Shock：490m/s２ Each direction 3 times
Dimension：110mm (Width) × 110mm (Height) × 104.5mm (Depth) Body diameter：99mmφ
Construction
With terminal coverMaterial
Case，Cover：ABS(V-0) ，Terminal block：PBT ，Terminal cover：Polycarbonate
Color
Black (Munsell N1.5)
Mass
Approx. 520g
Warranty at blackout
Maximum value, Minimum value, Setting value. Nonvolatile memory in data holds.
Operation temperature and-10 to +55℃, 30 to 85％ RH Non condensing.
humidity range
Storage temperature range -25 to +70℃
43

XLC-219-056

7.4 Option specifications 〔Production of the following products with input and output is possible by designating.〕
Item

Specification
Nomber of output
circuits

AnalogoutputExternal
operation
input

Maximum of 3 circuit

DC4 to 20mA (Below 550Ω)
DC0 to 1mA (Below 10kΩ)DC1 to 5V(Over 600Ω)
Analog output designates either with rating ofRating
DC0 to 5V(Over 600Ω)
identification. (22)DC0 to 10V (Over 2kΩ)
Other
Accuracy±0.5％
Response timeLess than 1 second (The time that final steady value deliver to ±1％)
Output ripple
For output span. Less than 1％ P-P.
Mutual output interval：Non-insulation. (Minus common)
Input factor
External reset ×1
External reset：
By applying a voltage signal from the outside, the maximum value and the minimum
value are reset. An input is the same rating as an auxiliary power supply.
Input(1) AC100/110V 0.4VA, AC200/220V 1.4VA, DC100/110V 0.4W for both AC and DC uses
specification
Contact capacity：About 3mA (AC,DC100/110V), About 6mA (AC200/220V)(2) DC24V 0.3W, DC48V 1.2W
Contact capacity：About 10mA (DC24V), About 20mA (DC48V)

Note(22) 1) Analog output standard range ･･･ Voltage output：±100mV to ±10V
Current output：±500μA to +20mA，-10mA
The number of output circuits becomes the number of input
circuits, and uniformity.
2) Analog output special range ････ 1. In case all rating is not uniformity.
2. In case the number of inputs and the number of outputs are
not the same. (Example：3 input → 1 output)
3. Current output：±100μA to ±499μA(The analog output accuracy changes to ±0.5％→±1.0％)● Caution on the use of external operation input (Option)
Power consumption of external operation input is 0.4VA at AC110V or 1.4VA at
AC220V or 0.4W at DC110V.
When a relay or a switch is used for power feed, its minimum application load
should be about 1mA.

P

N
19
20

8. Trouble shooting
Symptoms
Indicator does not
display.
The error of measurement
value is large.
Analog output is not
outputted.

Possible causes
The power supply is not supplied.
(Not connected. or voltage is low)

Remedial measures
Check the auxiliary supply.
Reapply of auxiliary power.

Measurement display ON/OFF setting is set to OFF.

Please check setting

Trouble of products.

Replace the products.

Range setting is not right.

Please set again.

Analog output is set to OFF or measurement display
is set to OFF.

Please check setting

44

XLC-219-056
Attached chart 1.

Bar graph scale division details.
The number in a bar graph scale is equivalent to the number of the following figure liquid crystal screen.
(A number is not displayed on actual liquid crystal.)And, scale division changes with full-scale values.

1

2 3 4 5 6

7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

At fragment swing.
(0 to N)：LCD scale lighting.

0.4
0.8
1.6
2
3.2

0

0.8
1.6
3.2
4
6.4

1.2
2.4
4.8
6
9.6

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
0.4
0.5
0.8
1
1.6
2

0

0.8
1
1.6
2
3.2
4

1.2
1.5
2.4
3
4.8
6

1.6
2
3.2
4
6.4
8

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
0.6
1.2
1.5
2.4
3

0

1.2
2.4
3
4.8
6

1.8
3.6
4.5
7.2
9

Scale
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

Bar graph

0

0.2
0.5
1

0.4
1
2

0.6
1.5
3

0.8
2
4

1
2.5
5

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

0

0.5
1
2.5

1
2
5

1.5
3
7.5

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

45

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

XLC-219-056

At fragment swing.
(-N to 0)：LCD scale lighting.

-1.2
-2.4
-4.8
-6
-9.6

-0.8
-1.6
-3.2
-4
-6.4

-0.4
-0.8
-1.6
-2
-3.2

0

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
-1.6
-2
-3.2
-4
-6.4
-8

-1.2
-1.5
-2.4
-3
-4.8
-6

-0.8
-1
-1.6
2
-3.2
-4

-0.4
-0.5
-0.8
-1
-1.6
-2

0

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
-1.8
-3.6
-4.5
-7.2
-9

-1.2
-2.4
-3
-4.8
-6

-0.6
-1.2
-1.5
-2.4
-3

0

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
-1
-2.5
-5

-0.8
-2
-4

-0.6
-1.5
-3

-0.4
-1
-2

-0.2
-0.5
-1

0

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
-1.5
-3
-7.5

-1
-2
-5

-0.5
-1
-2.5

0

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

46

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple ofｎ
10. (10 )Scale value
is integral
multiple ofｎ
10. (10 )XLC-219-056

At both swing.
(-N to 0 to N)：LCD scale lighting.

-1
-1.6
-2
-3.2
-4
-6.4
-8

-0.5
-0.8
-1
-1.6
-2
-3.2
-4

0.5
0.8
1
1.6
2
3.2
4

0

1
1.6
2
3.2
4
6.4
8

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30
-1.2
-1.5
-1.8
-2.4
-3
-3.6
-4.5
-4.8
-6
-7.2
-7.5
-9
-9.6

-0.4
-0.5
-0.6
-0.8
-1
-1.2
-1.5
-1.6
-2
-2.4
-2.5
-3
-3.2

-0.8
-1
-1.2
-1.6
-2
-2.4
-3
-3.2
-4
-4.8
-5
-6
-6.4

0.8
1
1.2
1.6
2
2.4
3
3.2
4
4.8
5
6
6.4

0.4
0.5
0.6
0.8
1
1.2
1.5
1.6
2
2.4
2.5
3
3.2

0

1.2
1.5
1.8
2.4
3
3.6
4.5
4.8
6
7.2
7.5
9
9.6

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

-2.5
-5

-1.5
-3

-0.5
-1

0

1.5
3

0.5
1

2.5
5

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)

Scale value
is integral
multiple of
10. (10ｎ)Frequency (Hz)：LCD scale lighting.

45
55
45

55
60
50

65
65
55

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

Power factor (COS )

LEAD
0
0.5

0.4
0.7

LAG
0.8
0.9

1
1

0.8
0.9

：LCD scale lighting.

0.4
0.7

0
0.5

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

47

XLC-219-056

Reactive power (var)：LCD scale lighting.

LEAD
1
1.6
2
3.2
4
6.4
8

LAG

0.5
0.8
1
1.6
2
3.2
4

0.5
0.8
1
1.6
2
3.2
4

0

1
1.6
2
3.2
4
6.4
8

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

LEAD
1.2
1.5
1.8
2.4
3
3.6
4.5
4.8
6
7.2
7.5
9
9.6

0.8
1
1.2
1.6
2
2.4
3
3.2
4
4.8
5
6
6.4

LAG
0.4
0.5
0.6
0.8
1
1.2
1.5
1.6
2
2.4
2.5
3
3.2

0.4
0.5
0.6
0.8
1
1.2
1.5
1.6
2
2.4
2.5
3
3.2

0

0.8
1
1.2
1.6
2
2.4
3
3.2
4
4.8
5
6
6.4

1.2
1.5
1.8
2.4
3
3.6
4.5
4.8
6
7.2
7.5
9
9.6

Scale
Bar graph

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

LEAD
2.5
5

1.5
3

Scale value
is integral
multiple of
10. (10ｎ)

LAG
0.5
1

0

0.5
1

1.5
3

2.5
5

Scale
Bar graph

Scale value
is integral
multiple of
10. (10ｎ)

1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30

48

Scale value
is integral
multiple of
10. (10ｎ)

XLC-219-056

Head Office: 11-13 Hitotsuya 1-chome, Adachi-ku, Tokyo, 121-8639 Japan.
Phone：03-3885-2411 , Fax：03-3858-3966

Kyoto Office : 1-19 Ichinobe-Nishikawahara, Jyoyou-shi, Kyoto, 610-0114 Japan.
Phone：0774-55-1391 , Fax：0774-54-1353
DATE：May 21, 201049