XLC-110/XLC-110L

POWER LINE DC RECEIVING METEROVERVIEW●Display from electrical signals, which are conversed from measurement of physical elements.
●Display 3 measuring elements at the same time. Combine with the system and monitor centrally
via analog output or communication output.

FEATURES●In 1 unit, possible to measure DC inputs of 3 circuits at the same time.
●Insulation: 2000V between inputs; 2000V between input and output.
●Possible to set the full scale of measuring value optionally.
●Possible to select from 18 kinds of units and set them display on LCD screen.
●Possible to display 1 bar graph and digital display of 3 measuring elements at the same time.
●Analog output (Max. 3 outputs) and Communication output are available.
●Possible to maintain Maximum and Minimum of measuring values.
●Use high quality, high contrast of LCD for display.
●Use high luminance of white backlight (in case of "with Backlight" option).
Possible to select light-on, light-off and brightness when setting.

XLC-110 / 110L
110×110×105mm (520g)

TYPE NAME & SPECIFICATION CODE
1. Without BacklightType①XLC-110 -Specification Code

②

③

④

⑤

⑥

②Hard Model③Input Circuit④Input Range⑤Auxiliary Power⑥Reset Input(DI)

A No Backlight

1 DC1～5V

8 3 circuits

3 DC0～5V

2 DC20～56V

4 DC0～10V

Z Other than above

2. With Backlight①XLC-110L -D

1⑧LCD View Angle

0 None0 Upper view

Analog Output
2 0～1mA

5 DC4～20mA

3 1～5V

6 DC0～1mA

4 0～5V

7 DC0～5mA
8 DC0～10mA

5 0～10V
Communication Output

9 DC0～16mA

M Modbus RTU

A DC0～20mA

A Protocol A (Com. Output)

Z Other than above(1)Z Other than above(2)

③

④

⑤

⑥

②Hard Model③Input Circuit④Input Range⑤Auxiliary Power⑥Reset Input(DI)

With Backlight
WHITE

6 1 circuit

1 DC1～5V

7 2 circuits

2 DC0～1V
3 DC0～5V

4 DC0～10V
5 DC4～20mA⑧

０

1 4～20mA②

8 3 circuits⑦⑦Analog Output (3) /Communication Output

0 NoneAC85～253V &DC80～143V Dual use 2 With DI x1

6 1 circuit
7 2 circuits

2 DC0～1V--⑦⑦Analog Output (3) /Communication Output

0 NoneAC85～253V &DC80～143V Dual use 2 With DI x1
2 DC20～56V

0 NoneAnalog Output

Z Other than above

2 0～1mA

1⑧

０⑧LCD View Angle
F

Upper & Lower
Co-viewing angle

1 4～20mA
3 1～5V

6 DC0～1mA
7 DC0～5mA
8 DC0～10mA

4 0～5V
5 0～10V
Communication Output

9 DC0～16mA
A DC0～20mA

M Modbus RTU
A Protocol A (Com. Output)

Z Other than above(1)Z Other than above(2)Note(1) Manufacturing ranges of Input:
Standard range

1. Voltage Input: ±50mV～±300V
2. Current Input: ±500μA～±50mASpecial range

1. Input is not the same as Rated Input
2. Current Input from ±100μA～±499μA(Digital display accuracy will change from ±1.0% to ±1.5%)(2) Manufacturing ranges of Analog Output:
Standard range

1. Voltage Output: ±100mV～±10V
2. Current Output: ±500μA～±20mA, -10mA
3. Quantity of Output circuit is the same as Input circuit

Special range

1. Output is not the same as Rated Output
2. Quantity of Input is different 0utput (Ex. 3-Input→1-Output)
3. Current Output from ±100μA～±499μA(Digital display accuracy will change from ±0.5% to ±1.0%)(3) In the case that quantity of Analog Output and Input are the same.

1

XLC-110/XLC-110L■Equipment Specification
Connection

LCD Display
Time of Display update
Measurement
Operating Temperature/ Humidity
Storage Temperature
Material
Weight■Communication
Specification
Dimension■Auxiliary Power Specification
Power Consumption(With Backlight)

Power Consumption (NoBacklight)

Inrush Current(For With backlight & No backlightdual use)

POWER LINE DC RECEIVING METER

Input, Aux. power parts: by M4 Screw
Reset Input (DI), Output parts: by M3 Screw
Main monitor: Text height 10mm 4 digits
Sub-monitor (L): Text height 6mm 4 digits
Sub-monitor (R): Text height 6mm 4 digits
Bar graph: 30 dots
Approx. 1 sec (Bar graph: approx.0.25sec)
DC Input 3 circuits-10 to +55℃
30 to 85% (No condensation)-25 to +70℃
ABS(V-0)
Exterior Color: Black (Munsell N1.5)520g
Refer to outline drawing (Compatible with wide angle analog meter)

AC85～253V
50/60Hz
DC80～143V
DC20～56V
AC85～253V
50/60Hz
DC80～143V
DC20～56V
AC110V
AC220V
DC110V
DC24V
DC48V

12VA
6W
7W
10VA
5W
6W
5.2A (approx. 1.7ms)
10.4A (approx. 1.7ms)
3.7A (approx. 1.7ms)
5.5A (approx. 3.6ms)
10.9A (approx. 3.6ms)■Output Specification
・Analog Output

Quantity: Maximum 3 Outputs

4～20mA: 0-550Ω, 0～1mA: 0-10kΩ
1～5V: 600Ω or more, 0～5V: 600Ω or more, 0～10V: 2kΩ or more
(Select one of above)
Non-insulation (minus common) between analog outputs
Below 1 sec. Time to be within ±1% of final constant value
Under 1% p-p of output span

Rated Output
Response Time
Output Ripple■Input Specification

Input

DC1～5V
DC0～1V
DC0～5V
DC0～10V
DC4～20mA
DC0～1mA
DC0～5mA
DC0～10mA
DC0～16mA
DC0～20mA

Input
Specification

Approx. 1MΩ
Approx. 50Ω
Approx. 1kΩ
Approx. 200ΩApprox. 100ΩDI
(Max/ Min
Value)

Approx. 50Ω

■Communication Specification
・Protocol A

Communication method
Transmission Speed
Transmission CodeStart bit
Data length
ParityStop bit
Cable length
Address
Quantity of connection
Transmission character

TIA-485-A Half-duplex 2-wire,
Asynchronous communication
1200 / 2400 / 4800 / 9600 bps
NRZ
1 bit
7/8 bits
None / even / odd
1bit / 2 bits
1000m (Fully extended)
1～254
Can connect up to 31 units.
In case of over 32 units, please use repeater(connect up to 254 units)
ASCII CodePower
Consumption

Contact
Capacity

DI: Possible to reset Max/Min value when adding voltage signal; Input andAux. power are the same value; Min. pulse width is 300ms continuation.
AC, DC100/110V
0.4VA, 0.4W
AC200/220V
1.4VA
DC24V
0.3W
DC48V
1.2W
AC, DC100/110V
3mA
AC200/220V
6mA
DC24V
10mA
DC48V
20mA

・Modbus RTU
Communication method
Transmission Speed
Transmission CodeStart bit
Data length
ParityStop bit
Cable length
Address
Quantity of connection
Transmission code
Error detection

TIA-485-A Half-duplex 2-wire,
Asynchronous communication
4800 / 9600 / 19200 / 38400 bps
NRZ
1 bit
8 bitsNone / even / odd
1bit / 2 bits
1000m (Fully extended)
1～247
Can connect up to 31 units.
In case of over 32 units, please use repeater(connect up to 254 units)
BinaryCRC-16 (x16+x15+x2+1)

2

XLC-110/XLC-110L

POWER LINE DC RECEIVING METER■PERFORMANCE
Items

Digital
DisplayStandard
Display Range
Power Factor (cosφ)
DisplayFrequency Display
Reactive Power
Display (LEAD,LAG)
Maximum Scale

Bar graph
Display

Power Factor (cosφ)
DisplayFrequency Display

Reactive Power
Display (LEAD,LAG)
Bar graph Display Accuracy
Temperature EffectTime of Display update
Main monitor
Display
Sub-monitor (L)Setting
Sub-monitor (R)Element
Bar graph
Up-Low co-viewing
LCD view
Upper view
Warranty when power cut-off

Accuracy
Reference
Display
OutputJIS C 1102-1,2,7,9 : 1997 JIS C 1111-1989 JIS C 1010-1 : 1998 TIA-485-A : 2003
-9999～9999
±1.0%
±0.5%
Any setting of number of digits & decimal point location is possible(1) LEAD 0.500～1.000～LAG 0.500
±1.0%
±0.5%
4 digits and decimal point location are fixed(2) LEAD 0.000～1.000～LAG 0.000(1) 45.0～55.0Hz or 45.00～55.00Hz
3 digits or 4 digits are fixed.
(2) 55.0～65.0Hz or 55.00～65.00Hz
±1.0%
±0.5%
decimal point location are fixed(3) 45.0～65.0Hz or 45.00～65.00Hz
Any setting of number of digits and decimal point location is
LEAD 9999～0～LAG 9999
±1.0%
±0.5%
possible
10 Integer Multiple (10n) of 1, 1.2, 1.5, 1.6, 1.8, 2, 2.4, 2.5, 3, 3.2,
Note: -9900 ≦ N ≦ 9900 (Scale value x 10n = N)
3.6, 4, 4.5, 4.8, 5, 6, 6.4, 7.2, 7.5, 8, 9, 9.6(1) LEAD 0.5～1～LAG 0.5
Scale value is fixed(2) LEAD 0～1～LAG 0
LEAD, LAG will display when power factor is selected.
(1) 45～55Hz(2) 55～65Hz
Scale value is fixed(3) 45～65Hz
LEAD □～0～LAG □Range: LEAD 9900～0～LAG 9900
□ are the same as maximum value
LEAD, LAG will display when power factor is selected.
±5.0% (% against span)
Accuracy will not change when in 23℃±10℃
Approx. 1 sec (Approx. 0.25 sec for bar graph)
Element of Input 1～Input 3 (depend on Display pattern)
Element of Input 1～Input 3 (depend on Display pattern)
Element of Input 1～Input 3 (depend on Display pattern)
Element of Input 1～Input 3
75° view for each Up, Low, Left, Right
10° view for Up, 60° for Low, 60° for each Left, Right
Remain Max, Min and setting value
Measuring Range / Display Specification■UNIT DISPLAY
LCD only displays below 18 kinds of unit. If unit is not in this 18 kinds, we will use unit seal to stick on display screen.
LCD Display Units (18 Kinds)
Sticky seal Units (55 Kinds) (5)Main monitor
Sub-monitor
(1)
A
A
(1)
APm
(19)
L/h
(37)
Nm3/min
(2)
kA
kA
(2)
bar
(20)
L/min
(38)
N/m2
(3)
V
V
(3)
cm
(21)
mA
(39)
N/mm2
(4)
kV
kV
(4)
cosφ
(22)
mg/L
(40)
OPm
(5)
W
(5)
ELm
(23)
min-1
(41)
Pa
(6)
kW
(6)
Hz
(24)
mL/min
(42)
pH
(7)
MW
(7)
J
(25)
mm
(43)
ppm
(8)
℃
℃
(8)
K
(26)
m/h (4)
(44)
R
(9)
%
%
(9)
kg
(27)
m/min (4)
(45)
rad
(10)
m
m
(10)
kg/h
(28)
m/s
(46)
rpm
(11)
m3
m3
(11)
kg/m2
(29)
mV
(47)
SPm
(12)
m3/h
m3/h
(12)
kg/m3
(30)
m3/s
(48)
t
(13)
m3/minm3/min
(13)
kL
(31)
MPa
(49)
t/h
(14)
m/h
(14)
kN
(32)
Mvar
(50)
TPm
(15)
m/min
(15)
kPa
(33)
MW (4)
(51)
W (4)
(16)
r/min
r/min
(16)
kvar
(34)
N
(52)
YPm
(17)
min
min
(17)
kW (4)
(35)
N・m
(53)
μm
(18)
no display
no display
(18)
L
(36)
Nm3/h
(54)
μS/cm

(55)

°C

Note: (4) Possible for sticky seal unit only on sub-monitors. Main monitor will display by LCD.
(5) Letter height of sticky seal: 8.5mm in main monitor, 5mm in sub-monitor. Letter’s color: gray (DIC 13th541). Units of sticky seal are selected when ordering, and cannot be changed after purchased.

■ DIMENSIONS (Unit: mm)3

XLC-110/XLC-110L

POWER LINE DC RECEIVING METER■ CONNECTION DIAGRAM
1-INPUT

2-INPUT

3-INPUTCOMMUNICATION OUTPUT

Note: (6) Input 1, Input 2, Input 3, DI (Reset Input), Pulse Output, Communication Output are options (selected by customer)(7) Terminating resistance can be connected internally by short circuit between Terminal (-) 14 and 16 (Ter.).

■ ORDER INFORMATION
Please specify below information to order:① Type name, Specification code
② Display Scales, Units
Please advise us the display scale and unit of each Input.
For Ex.: Input 1: 100.0℃ Input 2: 60.0℃ Input 3: 60.0℃③ Display Patterns
Please refer page 10 and advise us the patterns (Display form) from 1 to 6.

PARTS NAME & FUNCTIONBAR GRAPH DISPLAY
Display measuring values.
The measuring elements displayed on bar graphcan be changed by DISPLAY
button.

DIGITAL DISPLAY
3 elements can be measured and monitored at the same time.
Sub-monitor (Left) Main monitor Sub-monitor (Right)
SCALE MARKING
Scale display can be set automatically

FLICKER SETTING INDEXCan use Flicker OFF setting to manage the index.

UNIT DISPLAY
Can select any units in unit setting
DISPLAYFor changing the input element which will display on bar
graph

SET

Main monitor

For selecting the setting mode.
Display will change to setting mode 1 whencontinuously press ON more than 3sec.
This button is used for value setting.

Sub-monitor(Left)

Sub-monitor(Right)Display mode will change to setting mode 2 if
continuously pressing together with SET button in morethan 3sec. The function can be replaced by + button.
Setting mode will return to display mode after 1 action.

－
For Flicker value confirmation.
If no operation in 10sec, Flicker value display will
return to display mode. This button is also used
to carry down the setting value in setting mode.

+

For Max.value and Min.value confirmation in each input. Itchanges the order from the Max.value to Min.value. If no
operation in 10min, Max. and Min.value will return to display
mode. This button is used for carring up the setting value. It
can replace the function with DISPLAY button by setting.

RESET/SHIFTFor resetting the saved Max. and Min.value.
When continuously pressing ON more than 1sec, Max. andMin.value will be reset.
Use this button to move the items in Setting mode.

4

XLC-110/XLC-110L

POWER LINE DC RECEIVING METERSETTINGSETTING - 1

SETTING - 2

SETMeasuring Display ModeDISPLAYRESET/SHIFT & (+)

DISPLAYRESET/SHIFT & (-)
DISPLAY8

Note: ( ) Setting only for Backlight specification

Setting No. 231～233

Communication Output specification

DISPLAYRESET/SHIFT & (-)

Setting No. 271～273
Fix ed band of Display setting
RESET/SHIFT & (-)

Setting No. 281～283
Measuring Display ON/OFF setting
RESET/SHIFT & (+)

Note: Setting mode is a little different in case of

RESET/SHIFT & (-)

Setting No. 261～263
Low input cut setting

RESET/SHIFT & (+)
DISPLAYRESET/SHIFT & (-)

Setting No. 251b～256F
Calibration Output setting (9 )RESET/SHIFT & (+)
DISPLAYRESET/SHIFT & (-)

Setting No. 241A～243A
Output elements setting (9 )RESET/SHIFT & (+)
DISPLAYRESET/SHIFT & (-)

Bar graph Scale setting

RESET/SHIFT & (+)
DISPLAYRESET/SHIFT & (-)

Setting No. 221b～226F
Calibration Input setting

RESET/SHIFT & (+)
DISPLAYRESET/SHIFT & (-)

Setting No. 211b～21c
Display Scale setting

RESET/SHIFT & (+)

RESET/SHIFT & (-)

Setting No. 151～152 (8 )Backlight setting

RESET/SHIFT & (+)

RESET/SHIFT & (+)

RESET/SHIFT & (-)

Setting No. 131
DISPLAY button
Function change setting
RESET/SHIFT & (+)

DISPLAYDISPLAY

Setting No. 121H～129
Flicker setting
RESET/SHIFT & (+)

DISPLAY

3 sec

Setting No. 111～115
Pattern and Unit setting
RESET/SHIFT & (+)

DISPLAYRESET/SHIFT & (-)

Measuring Display Mode

3 sec

SET & DISPLAYRESET/SHIFT & (-)

Setting No. 291
Return to Defaut setting value

Note: (16 ) There is no output in case that Analog Output is not selected5

TLC-110/TLC-110L

DC MULTI METEROVERVIEW●Possible to monitor 3 measurements of Voltage (V), Current (A), Power (W), Energy (Wh) of DC
circuits at the same time.
●Possible to combine with the system and monitor centrally via analog output or communication output.

FEATURE●In 1 unit, possible to display 3 measurement at the same time.
●Possible to set your optional value on Full scale.
●By DC Current Input & DC Voltage Input, Power (W, Wh) will be calculated to output and display.
TLC-110 / 110L
●Insulation: 2000V between inputs; 2000V between input and output.
110×110×105mm (520g)●Possible to check the display via bar graph like an analog meter.
●Possible to maintain Max.value and Min.value.
●High-Low Flicker setting is available (with setting index). Can use Flicker OFF setting to manage the index.
●3 Analog Outputs or 2 Analog Outputs + 1 Pulse Output and communication output + 1 Pulse Output are available (option).
●Digital Input (DI – Reset Max.value & Min.value) is available.
●Wide range of Aux. power: AC85-253V, DC80-143V are dual use; DC20-56V is also available.
●Possible to replace our 110mm size Analog meter due to same form of installation on the panel (Installed by 2 screws).
●Use high quality, high contrast of LCD for display.
●Use high luminance of white backlight (in case of "with Backlight" option). Possible to select light-on, light-off and brightness setting.

TYPE NAME & SPECIFICATION CODE
1. Without BacklightType①TLC-110-Specification Code②③

②Hard Model
No
Backlight

DC Input Measuring

A

1

DC W・Wh Measuring()④③Input 1④Input 2

No Input
DC0～50mV
DC0～60mV
DC0～100mV
DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC0～1mA
DC±301～Y
4±600V( )No Input
DC0～50mV
DC0～60mV
DC0～100mV
DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC0～1mA
DC±301～Y
4±600V( )Current (A)Voltage (V)DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC±301～Y
4±600V( )

0
1
2
3
4
5
6
7
8
9

0
1
2
3
4
5
6
7
8
9

⑤

⑥

⑦

1 DC0～50mV
2 DC0～60mV
3 DC0～100mV

4
5
6
7
8

W

③

④

⑤

⑥

⑦

③Input 1④Input 2⑤Input 3⑥Auxiliary PowerWith
D Backlight
WHITE

No Input
DC0～50mV
DC0～60mV
DC0～100mV
DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC0～1mA
DC±301～Y
4±600V( )No Input
DC0～50mV
DC0～60mV
DC0～100mV
DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC0～1mA
DC±301～Y
4±600V( )No Input
DC0～50mV
DC0～60mV
DC0～100mV
DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC0～1mA
DC±301～Y
4±600V( )AC85～253V
1 DC80～143V
dual use
2 DC20～56V

Current (A)1 DC0～50mV
2 DC0～60mV
3 DC0～100mV

Voltage (V)
4 DC0～50V
5 DC0～75V
6 DC0～100V
7 DC0～150V
8 DC0～200V
DC±301～Y
4±600V( )

W・Wh
DC W
W
DC Wh

1

DC W・Wh Measuring⑩⑩LCD View
Angle

A

Z

0 Upper view

Other thanabove

Protocol A(Com. Output)

Z Other than above

W・Wh
DC W
DC Wh

②Hard Model()⑨⑨PulseOutput0 None1
output1(a contact)

Z Other than above Z Other than above②

DC Input Measuring-⑧⑧Analog Output /Communication Output
0 NoneAnalog Output
1 4〜20mA
2 0〜1mA
3 1〜5V
4 0〜5V
5 0〜10V
Communication Output
M Modbus RTU

Z Other than above Z Other than above Z Other than above

2. With Backlight①TLC-110L-⑦Reset⑤Input 3⑥Auxiliary Power(DI)
0 No Input
AC85～253V 0 None
1 DC0～50mV
1 DC80～143V 2 with DIx1
2 DC0～60mV
dual use
3 DC0～100mV 2 DC20～56V
Z Other than above
4 DC0～50V
5 DC0～75V
6 DC0～100V
7 DC0～150V
8 DC0～200V
9 DC0～1mA
DC±301～Y
4±600V( )

0
1
2
3
4
5
6
7
8
9

0
1
2
3
4
5
6
7
8
9

0
1
2
3
4
5
6
7
8
9

⑦Reset(DI)

0 None2 with DIx1

Z Other than above⑧Analog Output /Communication Output

0 NoneAnalog Output
1 4〜20mA
2 0〜1mA
3 1〜5V
4 0〜5V
5 0〜10V
Communication Output
M Modbus RTU⑨⑩⑨PulseOutput⑩LCD View
Angle

0 None1Z

1 output(a contact)0Upper & Lower
Co-viewing

Other thanabove

Protocol A(Com. Output)

Z Other than above(2) Manufacturing range of DC Input:
Standard range
Special range

1. Current Input: ±500μA～±50mA
2. Voltage Input: ±50mV～±600V
1. Current Input from ±100μA～±499μA(Digital display accuracy will change from ±1.0% to ±1.5%)
2. Voltage Input is from ±601V～±800V( ) Manufacturing range of Input for W, Wh measurement
3Standard range

Note:(1)・For W, Wh Measuring, please select specification of Current (code 1～3) for Input 1,
Voltage (code 4～8, Y)for Input 2, select “W” for Input 3.
・For display form of W and Wh, please select pattern 7～J.
・Current input is received from Shunt (we sell Shunt separately). In case of input not from Shunt,or±input, please select code “Z”.

⑧

A

Z Other than above Z Other than above Z Other than above

Z Other than above Z Other than above-

Special range

1. Current Input: ±50mV～±10V (output of Shunt, hole CT,…)
2. Voltage Input: ±5V～±600V
1. .Current Input: not the same as Rated Input
2. Voltage Input: ±601V～±800V

In case that Current Input is over ±50mA, please use with Shunt (purchase separately).
For Ex.: If Shunt is 50A/60mV, please use code 2 for Input 1.
(4) If Voltage Input is over ±301V, resistor DM-1 (accessory) will be attached (no fee caused).

XLC-110/110L TLC-110/110L Catalog | 5.0 en | efc2754

6

TLC-110/TLC-110L

DC MULTI METER■Auxiliary Power Specification■Equipment Specification
Connection

LCD Display
Time of Display update
Measurement
Operating Temperature/
Humidity
Storage Temperature
Material
WeightDimension

Input, Aux. power parts: by M4 Screw
Reset Input (DI), Output parts: by M3 Screw
Main monitor: Text height 10mm 4 digits
Sub-monitor (L): Text height 6mm 4 digits
Sub-monitor (R): Text height 6mm 4 digits
Bar graph: 30 dots
Approx. 1 sec (Bar graph: approx.0.25sec)
DC Input 3 circuits, W (calculated), Wh (calculated)-10 to +55℃
30 to 85% (No condensation)-25 to +70℃
ABS(V-0)
Exterior Color: Black (Munsell N1.5)520g
Refer to outline drawing (Compatible with wide angle analog meter)

Power
Consumption(With Backlight)
Power
Consumption(No Backlight)
Inrush Current(For backlight &no backlight dualuse)

AC85～253V
50/60Hz
DC80～143V
DC20～56V
AC85～253V
50/60Hz
DC80～143V
DC20～56V
AC110V
AC220V
DC110V
DC24V
DC48V

12VA
6W
7W
10VA
5W
6W
5.2A (approx. 1.7ms)
10.4A (approx. 1.7ms)
3.7A (approx. 1.7ms)
5.5A (approx. 3.6ms)
10.9A (approx. 3.6ms)■Input Specification

Input

DC0～50mV
DC0～60mV
DC0～100mV
DC0～50V
DC0～75V
DC0～100V
DC0～150V
DC0～200V
DC0～1mA

Input
Specification
DI (5)(Max/
Min Value)

Approx. 1MΩ

PowerConsumption
Contact
Capacity

Approx. 1kΩ

DI: Possible to reset Max/Min value when adding voltage signal; Input andAux. power are the same value; Min. pulse width is 300ms continuation.
AC, DC100V/110V
AC200V/220V
DC24V
DC48V
AC, DC100V/110V
AC200V/220V
DC24V
DC48V

0.4VA, 0.4W
1.4VA
0.3W
1.2W
3mA
6mA
10mA
20mA

Note (5) In DI (Reset Input), it is impossible to reset Power (Wh)■Output Specification
・Analog Output

Quantity: Maximum 3 Outputs
4～20mA: 0-550Ω, 0～1mA: 0-10kΩ,
Rated Output
1～5V: 600Ω or more, 0～5V: 600Ω or more, 0～10V: 2kΩ or more (Select one of above)
Non-insulation (minus common) between analog outputs
Response Time
Below 1 sec. Time to be within ±1% of final constant value
Output Ripple
Under 1% p-p of output span・Pulse Output
Output element: DC Power (Wh)Output Method
Photo MOS-FET relay 1a contact (N.O. contact)
Contact Capacity
AC, DC125V
250ms±10%
Pulse width
100～300ms depending on setting range
Unit of Pulse Output is possible to set by below range:
Multiplier
Unit of Pulse Output
0.1
1
0.1
0.01
0.001
1
10
1
0.1
0.01
10
100
10
1
0.1
100
1000
100
10
1
1000
(7) 10000
1000
100
10

Note: (6) In case of Pulse period speed setting is more than 1 pulse/sec
For Ex.
←If multiplier 1 isselected, units ofPulse output can
set as 10, 1, 0.1,
0.01■Communication Specification
・Protocol A

Communication method
Transmission Speed
Transmission CodeStart bit
Data length
ParityStop bit
Cable length
Address
Quantity of connection
Transmission character

Setting range of multiplier is limited as below due to Power scale (VxA)
Voltage (V) x Current (A)Possible range of multiplier＜ 100kW×0.1, ×1, ×10, ×100, ×1000
100kW ≦ ＜1000kW
×1, ×10, ×100, ×1000
1000kW ≦ ＜10000kW
×10, ×100, ×1000
10000kW ≦ ＜100000kW
×100, ×1000(𝑃𝑃𝑃𝑃𝑃𝑃𝑃𝑃𝑃𝑃𝑉𝑉𝑉𝑉𝑉𝑉
𝑜𝑜𝑜𝑜𝑜𝑜𝑜𝑜𝑜𝑜𝑜𝑜 𝑢𝑢𝑢𝑢𝑢𝑢𝑢𝑢 ≧3600), Pulse width is 100-130ms. It loads about 2Hz when
(𝑘𝑘𝑘𝑘)

Pulse width is 250ms, 4.5Hz when Pulse width is 100-130ms.
(0 < Pulse output (Hz) < 2Hz or 4.5Hz)(7) Because there are only 4 digits, unit 10000 of Pulse output will display: 9999

・Modbus RTU

TIA-485-A Half-duplex 2-wire,
Asynchronous communication
1200 / 2400 / 4800 / 9600 bps
NRZ
1 bit
7/8 bits
None / even / odd
1bit / 2 bits
1000m (Fully extended)
1～254
Can connect up to 31 units.
In case of over 32 units, please use repeater(connect up to 254 units)
ASCII CodeCommunication method
Transmission Speed
Transmission CodeStart bit
Data length
ParityStop bit
Cable length
Address
Quantity of connection
Transmission code
Error detection

XLC-110/110L TLC-110/110L Catalog | 5.0 en | efc2754

TIA-485-A Half-duplex 2-wire,
Asynchronous communication
4800 / 9600 / 19200 / 38400 bps
NRZ
1 bit
8 bitsNone / even / odd
1bit / 2 bits
1000m (Fully extended)
1～247
Can connect up to 31 units.
In case of over 32 units, please use repeater(connect up to 254 units)
BinaryCRC-16 (x16+x15+x2+1)

7

TLC-110/TLC-110L

DC MULTI METER■PERFORMANCE
Item

Accuracy
Reference
Display
OutputJIS C 1102-1,2,7,8,9 : 1997 JIS C 1111-1989 JIS C 1010-1 : 1998 TIA-485-A : 2003
-9999～9999
±1.0%
±0.5%
Any setting of number digits & decimal point location is possible
in Wh measurement: 0～9999, display up
±3.0%
±3.0%
When display is over 9999, it will continuously count from 0
to 3rd decimal digit.
n
10 Integer Multiple (10 ) of 1, 1.2, 1.5, 1.6,
% against span
1.8, 2, 2.4, 2.5, 3, 3.2, 3.6, 4, 4.5, 4.8, 5, 6, ±5.0%－
Note: -9900 ≦ N ≦ 9900 (Scale value x 10n = N)
6.4, 7.2, 7.5, 8, 9, 9.6
Accuracy will not change when in 23℃±10℃
Approx. 1 sec (Approx. 0.25 sec for bar graph)
Element of Input 1～Input 3 (When measuring W, elements: A, V, W. When measuring Wh, element: Wh)
Element of Input 1～Input 3 (When measuring W, elements: A, V, W)
Element of Input 1～Input 3 (When measuring W, elements: A, V, W)
Element of Input 1～Input 3 (When measuring W, elements: A, V, W)
75° view for each Up, Low , Left, Right
10° view for Up, 60° for Low, 60° for each Left, Right
Remain Max, Min, setting and calculation values
Measuring Range / Display Specification

Standard
Digital
Display

Display Range

Bar graph
Display

Maximum Scale

Temperature EffectTime of Display update
Main monitor
Display
Sub-monitor (L)Setting
Sub-monitor (R)Element
Bar graph
Up-Low co-viewing
LCD view
Upper view
Warranty when power cut-off■UNIT DISPLAY

LCD only displays below 18 kinds of unit. If unit is not in this 18 kinds, we will use unit seal to stick on display screen.
LCD Display Units (18 Kinds)
Sticky seal Units (56 Types) (5)Main monitor
Sub-monitor
(1)
A
A
(1)
APm
(19)
L/h
(37)
Nm3/min
(55)
°C
(2)
kA
kA
(2)
bar
(20)
L/min
(38)
N/m2
(56)
kWh
(3)
V
V
(3)
cm
(21)
mA
(39)
N/mm2
(4)
kV
kV
(4)
cosφ
(22)
mg/L
(40)
OPm
(5)
W
(5)
ELm
(23)
min-1
(41)
Pa
(6)
kW
(6)
Hz
(24)
mL/min
(42)
pH
(7)
MW
(7)
J
(25)
mm
(43)
ppm
(8)
℃
℃
(8)
K
(26)
m/h (8)
(44)
R
(9)
%
%
(9)
kg
(27)
m/min (8)
(45)
rad
(10)
m
m
(10)
kg/h
(28)
m/s
(46)
rpm
(11)
m3
m3
(11)
kg/m2
(29)
mV
(47)
SPm
(12)
m3/h
m3/h
(12)
kg/m3
(30)
m3/s
(48)
T
(13)
m3/minm3/min
(13)
kL
(31)
MPa
(49)
t/h
(14)
m/h
(14)
kN
(32)
Mvar
(50)
TPm
(15)
m/min
(15)
kPa
(33)
MW (8)
(51)
W (8)
(16)
r/min
r/min
(16)
kvar
(34)
N
(52)
YPm
(17)
min
min
(17)
kW (8)
(35)
N・m
(53)
μm
(18)
no display
no display
(18)
L
(36)
Nm3/h
(54)
μS/cm

Note: (8) Possible for sticky seal unit only on sub-monitors. Main monitor will display by LCD.
(9) Letter height of sticky seal: 8.5mm in main monitor, 5mm in sub-monitor. Letter’s color: gray (DIC 13th541). Units of sticky seal are selected when ordering, and cannot be changed after purchased.

■ DIMENSIONS (Unit: mm)

DM-1 Dimension

Attached to each Input whenvoltage input is over ±301V

XLC-110/110L TLC-110/110L Catalog | 5.0 en | efc2754

8

TLC-110/TLC-110L

DC MULTI METER■ CONNECTION DIAGRAM
1-INPUT

2-INPUT

3-INPUTFor W, Wh measurement,
shunt for connection ispurchased separately.
When Voltage input is over
301V, resistor (Type: DM-1)
is attached (no fee caused)
By Input 1 (A) & Input 2(V), Power (W, Wh) will
be calculated to output
and display

COMMUNICATION OUTPUT

Note: (10) Input 1, Input 2, Input 3, DI (Reset Input), Pulse Output, Communication Output are options (selected by customer).
(11) Output 1, 2 are Analog type, Output 3 is Analog type or Pulse type. (Terminal 17(+), 18(-) if Output 3 is Analog). Output of Wh is Pulse Output in Output 3.
(12) Terminating resistance can be connected internally by short circuit between Terminal (-) 14 and 16 (Ter.).

■ ORDER INFORMATION Please specify below information to order:
1. For measuring DC Input① Type name & Specification code
② Display Scales, Units: Please advise us the display scale and unit of each Input.
For Ex.: Input 1: 100.0A Input 2: 200.0V Input 3: 200.0V
2. For measuring W, Wh① Type name, Specification code
② Display Scales, Units: Please advise us the display scale and unit of each Input.
For Ex.: Input 1: 100.0A Input 2: 200.0V Input 3: 200.0kW③ Multiplier (In case of Wh): Please select at below table:
Voltage (V) x Current (A)＜ 100kW
100kW ≦ ＜1000kW
1000kW ≦ ＜10000kW
10000kW ≦ ＜100000kW

Possible range of multiplier×0.1, ×1, ×10, ×100, ×1000
×1, ×10, ×100, ×1000
×10, ×100, ×1000
×100, ×1000

Cannot change the multiplier after purchase.
Example of Order:
Type name & Specification code: TLC-110L-D24W12-11F
2-Input selected: Input 1: 2000A/60mV Input 2: 50.0V/50V W: 100.0kW Multiplier×10kWh
AC85-253V, DC80-143V
2-Output selected: DC4-20mA (below 550Ωload), Output 1: A, Output 2: V, Pulse Output: 10kWh/Pulse
With DI (Reset Input) Display Pattern: D Quantity of order: 1 pc(s)③ Analog output and quantity (1-3) of output.
④ Display Patterns:
Please refer page 10 and advise us the patterns (Display form) from 1 to 6.
④ Units of Pulse Output (In case of Pulse Output): Please select at below table:
Multiplier
0.1
1
10
100
1000

1
10
100
1000
10000

Unit of Pulse Output
0.1
0.01
1
0.1
10
1
100
10
1000
100

0.001
0.01
0.1
1
10

⑤ Analog output and quantity of output.
In case of “No Pulse Output”: please select quantity of Analog Output from 1-3
In case of “With Pulse Output”: please select quantity of Analog Output from 1-2⑥ Display Patterns: Please refer page 10 and advise us the patterns (Display form)
For measuring W: Select pattern from 7-C. For measuring Wh: Select pattern from D-J
Select multiplier for measuring Wh
Select Unit for Pulse Output

XLC-110/110L TLC-110/110L Catalog | 5.0 en | efc2754

9

TLC-110/TLC-110L

DC MULTI METERPARTS NAME & FUNCTIONBAR GRAPH DISPLAY
Display measuring values.
The measuring elements displayed on bar
graph can be changed by DISPLAY
button.

DIGITAL DISPLAY
3 elements can be measured and monitored at the same time.
Sub-monitor (Left)
Main monitor
Sub-monitor (Right)
SCALE MARKING
Scale display can be set automaticallyTLC-110L

FLICKER SETTING INDEXCan use Flicker OFF setting to manage theindex.

UNIT DISPLAY
Select units in unit setting
DISPLAYFor changing the input element which will display on
bar graph

SET
For selecting the setting mode.
Display will change to setting mode 1 whencontinuously press ON more than 3sec.
This button is used for value setting.
When measuring Wh, it can display up to the 3rd
decimal digit.

Main monitor－
For Flicker value confirmation.
If no operation in 10sec, Flicker value display will
return to display mode. This button is also used
to carry down the setting value in setting mode.

SETTINGSET
3 sec

DISPLAY+For Max.value and Min.value confirmation in each input. Itchanges the order from the Max.value to Min.vaue. If no
operation in 10min, Max. and Min.value will return to display
mode. This button is used for carry up the setting value. It canreplace the function with DISPLAY button by setting.

SETTING - 1
RESET/SHIFT & (+)

SET & DISPLAY
3 sec

RESET/SHIFT & (-)

Setting No. 111～115
Pattern and Unit setting

DISPLAY

Measuring Display Mode

Setting No. 121H～129
Flicker setting

DISPLAY

Setting No. 131
DISPLAY button
Function change setting

DISPLAY

Setting No. 14 ( )142P (14 ) multiplier
Unit of Pulse output setting
RESET/SHIFT & (+) RESET/SHIFT & (-)

DISPLAY

Note:

Setting No. 151～152(15 )Backlight setting

13

( )

Setting only for measuring Wh (Display Pattern D-J)

( )

14

Setting only for Pulse output (option)

15

Setting only for Backlight specification( )RESET/SHIFT& (-)

Setting No. 211b～21c
Display Scale setting

Setting No. 221b～226F
Calibration Input setting

Setting No. 231～233
Bar graph Scale setting
RESET/SHIFT & (+) RESET/SHIFT & (-)

Measuring Display Mode

13

RESET/SHIFT& (+)

RESET/SHIFT & (+) RESET/SHIFT & (-)

RESET/SHIFT & (+) RESET/SHIFT & (-)
DISPLAYFor resetting saved Max. and Min.value.
When continuously press ON more than 1sec, Max.
and Min.value will be reset.

RESET/SHIFT & (+) RESET/SHIFT & (-)

RESET/SHIFT & (+) RESET/SHIFT & (-)
DISPLAYRESET/SHIFTSETTING - 2

RESET/SHIFT & (+) RESET/SHIFT & (-)
DISPLAY

Sub-monitor(Right)Display mode will change to setting mode 2 if
continuously press together with SET button in morethan 3sec. The function can be replaced by +button. Setting mode will return to display mode after
1 action.

DISPLAY

SET

Sub-monitor(Left)DISPLAY

Setting No. 241A～243A
Output elements setting (16 )RESET/SHIFT & (+) RESET/SHIFT & (-)

DISPLAY

Setting No. 251b～256F
Calibration Output setting (16 )RESET/SHIFT & (+) RESET/SHIFT & (-)

DISPLAY

Setting No. 261～263
Low Input cut setting
RESET/SHIFT & (+) RESET/SHIFT & (-)

DISPLAY

Setting No. 271～273
Fix ed band of Display setting
RESET/SHIFT & (+) RESET/SHIFT & (-)

DISPLAY

Setting No. 281～283
Measuring Display ON/OFF setting
RESET/SHIFT & (+) RESET/SHIFT & (-)

Note: Setting mode is a little different in case of

DISPLAY

Communication Output specification

Setting No. 291
Return to Defaut setting value

16

Note: ( ) There is no Output in case that Analog Output is not selected

XLC-110/110L TLC-110/110L Catalog | 5.0 en | efc2754

10

COMMON SPECIFICATION■COMMON SPECIFICATION
Product Name
Type
Name

DC Receiving Meter

DC Multi Meter

No Backlight

XLC-110

TLC-110

With Backlight

XLC-110L

TLC-110L

Items

Specification
JIS C 1102-1 : 1997 Direct acting analog electrical measuring instruments. Part 1: Definitions and general requirements
JIS C 1102-2 : 1997 Direct acting analog electrical measuring instruments. Part 2: Special requirements for ammeters and voltmeters
JIS C 1102-7 : 1997 Direct acting analog electrical measuring instruments. Part 7: Special requirements for multi-function instruments

Standard

JIS C 1102-8 : 1997 Direct acting analog electrical measuring instruments. Part 8: Special requirements for accessories
JIS C 1102-9 : 1997 Direct acting analog electrical measuring instruments. Part 9: Recommended test methods
JIS C 1111 : 1989 AC-DC Transducer
JIS C 1010-1 : 1998 Safety requirements for electrical equipment for measurement, control, and laboratory use. Part 1: General requirements
TIA-485-A : 2003 Electrical characteristics of drivers and receivers for use in balanced and multipoint systems.

Withstand Overload

Voltage circuit

2 times of rated voltage for 10 sec., 1.2 times for continuous

Current circuit

10 times of rated current for 5 sec., 1.2 times for continuous

Auxiliary Power

1.5 times of rated voltage for 10 sec., 1.2 times for continuous
When aux. power is DC110V: 1.5 times of rated voltage for 10 sec., 1.3 times for continuous

Between electric circuit and the outside box (earth)
Between input, output and aux. power
Insulation Resistance

Between inputs

DC500V
50MΩ or more

Between outputs (analog / communication / pulse)
Between analog outputs

Non- isolation (minus common)

Between electric circuit and the outside box (earth)
Between input, output and aux. power
Withstand Voltage

AC2000V (50/60Hz), 1 min (1)Between inputs
Between outputs (analog / communication / pulse)

AC1500V (50/60Hz), 1 min

Between analog outputs

Non- isolation (minus common)

Lightning Impulse
Withstand Voltage

Between electric circuit and the outside box (earth)

5kV

Noise Capacity(1) Oscillatory surge voltage:
If the attenuated oscillatory waveform of 2.5-3kV peak voltage and 1-1.5MHz frequency is applied repeatedly, there is no effect tomeasurement and communication (Accuracy of measurement is within ±10%)
Voltage, Current input circuit (Common), Aux. power circuit (Normal/Common)(2) Square-wave impulse noise:
If the square-wave impulse noise (1μs, 100ns width) is applied repeatedly in 5 min, there is no effect to measurement and communication.
(Accuracy of measurement is within ±10%).
Voltage, current input circuit (Common): 1.5kV or more
Aux. power circuit (Normal/Common): 1.5kV or more
External input (DI) (Common): 1.0kV or more
Analog or Communication output circuit (Inductive): 1.0kV or more
Pulse output (Common): 1.0kV or more
(3) Radio noise:
If radio wave of 150,400MHz is applied continuously by 5W, 1m, there is no effect to measurement and communication.
(Accuracy of measurement is within ±10%)(4) Static noise:
When electric connection (noise voltage: 8kV): Accuracy of measurement is within ±10%
When no electric connection (noise voltage: 10kV): No damage caused. Condenser change method.

Vibration / ShockVibration: (Positive scale only) 0.15mm width, 10-55Hz: 5 times sweep in 1 octave/min
Shock: 490m2/s, X, Y, Z direction 3 times for each

Structure
Power Outage
Guarantee

1.2/50μs plus/minute polarity each 3 times

Dimension: length x height x width: 110 x 110 x 105mm. Diameter: 99mmΦ. Terminal cover attached. Protection class: IP40
Material of case: ABS(V-0). Out-looking color: Black (Munsell N1.5). Weight: approx.. 520g
Max. value, min. value, setting values, integrated values are saved by data memory

Operating Temperature/
Humidity Range-10℃～+55℃, 30～85% RH, no condensation

Storage Temperature-25～+70℃

Note: (1) When circuit voltage is 501~800V, withstand voltage is AC2200V.

11PATTERNS (DISPLAY FORMS)
XLC-110/110L
Specification of 1-input type
Main
SubSubmonitor monitor monitor(L)(R)
Input 1-

Specification of 2-input type
Main
SubSubmonitor monitor monitor(L)(R)
Input 1
Input 2
Input 1
Input 2
Input 2
Input 1
Input 2
Input 1
Input 1
Input 2
Input 2
Input 1

Specification of 3-input type
Main
SubSubmonitor monitor monitor(L)(R)
Input 1
Input 2
Input 3
Input 1
Input 3
Input 2
Input 2
Input 1
Input 3
Input 2
Input 3
Input 1
Input 3
Input 1
Input 2
Input 3
Input 2
Input 1

No.

Pattern
No.

1
2
3
4
5
6

Pattern 1
Pattern 2
Pattern 3
Pattern 4
Pattern 5
Pattern 6Pattern 1: Standard; Pattern 2~6: Specify and change the display position
Only display elements of Input circuit number (For Ex.: In case of 1-input, only display on main monitor (Input 1), sub-monitors are non-display.

Bar graph
Input 1 (can change to 2, 3)
Input 1 (can change to 2, 3)
Input 2 (can change to 1, 3)
Input 2 (can change to 1, 3)
Input 3 (can change to 1, 2)
Input 3 (can change to 1, 2)

INPUT
Display
ON/OFF

OFF
OFF
OFF
OFF
OFF

TLC-110/110L
Specification of 1-input type
SubSubMain
monitor monitor
monitor(L)(R)
Input 1-

Specification of 2-input type
SubSubMain
monitor monitor
monitor(L)(R)
Input 1
Input 2
Input 1
Input 2
Input 2
Input 1
Input 2
Input 1
Input 1
Input 2
Input 2
Input 1
W
A
V
W
V
A
W
A
V

Specification of 3-input type
SubSubMain
monitor monitor
monitor(L)(R)
Input 1
Input 2
Input 3
Input 1
Input 3
Input 2
Input 2
Input 1
Input 3
Input 2
Input 3
Input 1
Input 3
Input 1
Input 2
Input 3
Input 2
Input 1
W
A
V
W
V
A
W
A
V

No.

Pattern
No.

1
2
3
4
5
6
7
8

Pattern 1
Pattern 2
Pattern 3
Pattern 4
Pattern 5
Pattern 6
Pattern 7
Pattern 8

9Pattern 9

-

-

-

A

Pattern A

-

-

-

A

B

Pattern B

-

-

-

V

C

Pattern C

-

-

-

V

D

Pattern D

-

-

-

E

Pattern E

-

-

-

F

Pattern F

-

-

-

G

Pattern G

-

-

-

H

Pattern H

-

-

-

J

Pattern J

-

-

-



Only display elements of Input circuit number (For Ex.: In case of 1-input, only display on main monitor (Input 1), sub-monitors are non-display).
Depending on Patterns, it is possible to change position of display, and do the setting of DC (W), (Wh) measurements.
Pattern 1: Standard; Pattern 2~6: Specify and change the display position;
Pattern 7~C: Specify (W) measurement; Pattern D~J: Specify (Wh) measurement.
Units of sticky seal are selected when ordering (Ex. W, kW, Wh, kWh, etc.), and cannot be changed after purchased.

(sticky seal)Wh(sticky seal)Wh(sticky seal)Wh(sticky seal)Wh(sticky seal)W(sticky seal)A
W(sticky seal)

V
W(sticky seal)

A

A

V

V

A

A
V

Wh

W

(sticky seal)(sticky seal)Wh

W(sticky seal)(sticky seal)

V
V
Wh(sticky seal)Wh(sticky seal)W

Wh(sticky seal)(sticky seal)W

Wh(sticky seal)(sticky seal)

A
V

INPUT
Display

Input 1 (can change to 2, 3)
Input 1 (can change to 2, 3)
Input 2 (can change to 1, 3)
Input 2 (can change to 1, 3)
Input 3 (can change to 1, 2)
Input 3 (can change to 1, 2)
W (can change to A, V)
W (can change to A, V)

ON/OFF

A (can change to V, W)

OFF

A (can change to V, W)

OFF

V (can change to A, W)

OFFA

V (can change to A, W)

OFF

A

V

A (can change to V)

OFF

V

A

V (can change to A)OFF

A (can change to W)

OFF

V (can change to W)

OFFA

W (can change to A)

OFFV

W (can change to V)

OFF(sticky seal)A

Bar graphW(sticky seal)A
W(sticky seal)

A
V

Wh

W

(sticky seal)(sticky seal)Wh

W(sticky seal)(sticky seal)

V
W(sticky seal)W(sticky seal)W(sticky seal)

OFF
OFF
OFF
OFF
OFF
OFF
OFF

12

SAFETY PRECAUTIONSTo ensure safety, please follow carefully the below attentions. Manual instruction is enclosed in same box of product, please read it carefully before using and make surethat the manual instruction is read by end-user also.
 Usage environment and conditions
Please ensure to use this product in a place that meets the following conditions.
In places that do not meet this conditions, it may cause malfunction and reduce the product life.
Product is not in direct contact with rain, water drops or sunlight.
Within the ambient temperature range of -10 to +55℃.
Altitude is 2000m or lower.

Daily average temperature is not over 40℃.

If this product directly measures an inverter output of Cycle control,
Humidity is 85% RH or less, and non-condensing.
SCR phase angle control or PWM control, an error may occur dueFree of dust, corrosive gas, salt and oily smoke.
to its operation principle.
(Corrosive gas：SO2 / H2S, etc.)
 Outdoor use conditions
Please follow the below notices when using outside the panel:Please prevent this product from rain and water drops because this product is not waterproof and splash proof construction. (Protection class: IP40)Please avoid the place with much dust.

Please do not install in the place where sunlight hits directly even though product is behind glass.
Under the direct sunlight, screen of meter may get the high temperature and be deformed when over 80℃.

If the average temperature of surrounding area is over 40℃, the product life span may be decrease.
 Mounting
Please use 2.0~2.5N.m of torque to tighten the flange nuts M5 when mounting.

Please take care the LCD parts not to be shock to prevent the error and damage.
 Connecting

To ensure safety, connections should be performed by an electrical engineer qualified in wiring.

Please use crimping terminals to connect.
Please use 1.0~1.3N.m of torque for M4 screws and 0.5~0.6N.m of torque for M3 screws.
Terminal cover is for avoiding electric shock, please use terminal cover after conducting connection.
 Preparations before useThis product must be set primary voltage, etc… correctly before use. Incorrect setting may result in wrong display.
 Usage procedures
Use this product within the rated range because if out of rated range, it may cause erroneous operation or malfunction.

Please set the value correctly by reading Manual instruction carefully.
 Handling at Time of Malfunction/Error
If the product listed in this catalog malfunctions, cut off the power and input and stop using. Please contact Daiichi Electronics or our representatives.

In case of stripping down or remodeling, please be noticed that it may be out of warranty.
 About dew condensation
If the temperature and humidity change suddenly when this product is non-powered, the water drops by dew condensation may appear at inner side of display.
(The display filter may stick to LCD surface, and round or ellipse shape of patterns may appear)
This phenomenon does not cause any trouble. It will disappear when power supply is applied continuously for about 2 hours.
 Maintenance and inspections

Use a soft towel to clean the dirt on screen. If the dirt is too hard to clean, use a wet towel (squeezed by water with a little neutral soap).
Please do not use the organic solvent, chemicals, cleaners, etc., for cleaning.
Please inspect the product to check out the bellowing:(1) Damages on product(3) Screws on terminals are slack when mounting(2) Display is error or not (Ex. display does not respond to input) (4) Terminal parts have dust or not
Please cut off the power during inspection.
 Storage
In the long period of storage, please keep the product in a place that meets the following conditions.
In places that do not meet this conditions, it may cause malfunction and reduce the product life.

No shock and vibration.
Within the ambient temperature range of -20 to +70℃
Altitude is 2000m or lower.
(storage temperature).
Product is not in direct contact with rain, water drops or sunlight
Daily average temperature 40 ℃ or lower.
Free of dust, corrosive gas, salt and oily smoke.
During the storage period, do not connect with any wire of power, input/output and wrap the product in a vinyl pack.
 Disposal
This product do not use nickel-cadmium batteries. Dispose them as industrial waste (non-burnable trash).

DANGER OF
ELECTRIC SHOCK

Please refer to manual instruction for mounting and wiring, and should be performed by an electrical engineer.
 Please check the connection diagrams carefully before performing connections. Wrong connections may lead to machine faults
and burn-out.
 Please avoid working with live wires because it may result in electric shock, malfunctions, gas and fire.
 Terminal cover is for avoiding electric shock, please use terminal cover after performing connections.

13