TEST

Oppdragsgiver Client

Utførende enhet/lab. Department/laboratory responsible

Nortek AS
Vangkroken 21351 Rud

The National Institute of Technology
Environmental Test Laboratory
P.O. Box 1019
NO - 3601 KONGSBERG

Rapportnr. Report no.

3000-14-017757
Tittel Title

Environmental test report
Vibration test
Signature
S-55, S-500 and S-1000
Dato Date

01.07.2014

Utarbeidet av Prepared by

Godkjent av Approved by

Department 3000

Erik Haugejorden

Innleveringsdato for prøve
Date of receipt of test object

30.06.2014
Prøvetaking utført av TI
Sampling by TI

Nei NoRevisjonsnr. Revision no.

Konfig.kont. Config.contr .

-Antall sider No. of pages

8

Ja YesKopi nr. Copy no.

Ant. vedlegg No. of append.

Akkreditert test (ISO 17025)
Accredited test (ISO 17025)

Ja YesKundens ref. Client’s ref.

2Bestillingsnr. Order no.

Dag Skogbrott107580Fordeling Distribution

Dag Skogbrott, Nortek; 1 pdf-file.

Akkreditering innebærer at laboratoriet oppfyller de krav Norsk
Akkreditering stiller til kompetanse og prøvingssystemer for de prøvinger
akkrediteringen omfatter. Dette innebærer også at laboratoriet har ettilfredsstillende kvalitetssikringssystem og nødvendig sporbarhet til
akkrediterte eller nasjonale laboratorier. Prøvingsresultatene gjelder
ute lukkende de prøvede objekter. Utdrag av rapporten må ikke gjengis
uten skriftlig godkjenning fra Teknologisk Institutt as.

SK-0016 d – N/E

Postadresse:
Pb. 141 Økern0509 Oslo
Besøksadresse:
Kabelgaten 2, 0580 Oslo

Accreditation implies that the laboratory meets the requirements set
by Norwegian Accreditation concerning competence and testing system
for all the testing contained in the accreditation. It also states thatthe laboratory has a satisfactory quality assurance system andtraceability to accredited or national laboratories. The results relate
only to the items tested. The report shall not be reproduced except infull, without the written approval of the institute.

Teknologisk Institutt as
Telefon:Telefaks:
E-post:Web:

22 86 50 00
22 72 45 02
firmapost@ti.no
www.teknologisk.no

Org.nr. 942 340 680

Kongsberg:
Stavanger:
Haugesund:Bergen:Pb. 1019, NO-3601 Kongsberg
Forusbeen 10, NO-4033 Stavanger
Pb. 93, NO-4299 Avaldsnes
Pb. 23, NO-5346 Ågotnes

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

2 av of 8
Dato Date 01.07.2014

Contents
1.
2.
3.
4.
5.

Scope .......................................................................................................................................................... 3
Reference Documents ................................................................................................................................ 3
Test Procedure............................................................................................................................................ 3
Results ........................................................................................................................................................ 4Conclusion ................................................................................................................................................. 4

Appendix 1: Test Equipment.
Appendix 2: Test Records.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

3 av of 8
Dato Date 01.07.2014

1. Scope
The Environmental Test Laboratory at Teknologisk Institutt as has performed vibration test on Signature S55 / S/N S-0011, S-500 / S/N D-0069 and S-1000 / S/N D-0086 for Nortek AS. The test was performed
30.06.2014.
The test was executed under accreditation granted by Norwegian Accreditation, accreditation no. Test 006,
and was based on the general test requirements specified in IEC 60068-1 / IEC 60068-2-64.
The following personnel were present during the test:
Erik Haugejorden, Teknologisk Institutt AS.
Dag Skogbrott, Nortek AS.
Bengt Large, Nortek AS

2. Reference Documents
1.
2.
3.
4.

KS Handbook, Quality Manual for Teknologisk Institutt.
Environmental test report: 3010-09-0048, Teknologisk Institutt AS.
IEC 60068-1 General requirements, edition 7.0.
IEC 60068-2-64 Test Fh: Vibration, broad band random and guidance, edition 2.0.

3. Test Procedure
The environmental test equipment used during test with calibration status and measurement uncertainty
figures is given in Appendix 1. The test items were mounted on the shaker as shown in figures 1-3. The test
items were not powered during test. A visual inspection was performed after test. Functional test wasperformed at Nortek after test. The positions of control and response accelerometers are shown in figures 14.
MP1: Control accelerometer.
MP2: Response on S-55.
MP3: Response on S-1000.
MP4: Response on S-55.
The following test parameters were used:
Frequency range:
10 - 500Hz.
Excitation level:
10 - 200 Hz: 0.03 g2/Hz.
200 - 500 Hz: 0.01 g2/Hz.
Grms:
2.95
Duration:
30 minutes in each of the 3 axes.
The recorded input and response records are given in Appendix 2.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

4 av of 8
Dato Date 01.07.2014

4. Results
The test items were not operational during test. Mechanical inspection and functional tests were performed at
Nortek after the test. Test results are evaluated by Nortek.

5. Conclusion
The test items were not operational during test. Mechanical inspection and functional tests were performed at
Nortek after the test. Test results are evaluated by Nortek.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

5 av of 8
Dato Date 01.07.2014

MP 4
MP 3

1: Control

MP 2

Figure 1.
Test items mounted for vibration test in x-axis.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

6 av of 8
Dato Date 01.07.2014MP 4
1: Control

MP 3

MP 2

Figure 2.
Test items mounted for vibration test in y-axis.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

7 av of 8
Dato Date 01.07.2014

MP 2

1:Control.

Figure 3.
S-55 mounted for vibration test in z-axis.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Side Page

8 av of 8
Dato Date 01.07.2014

MP 3

MP 4

1:Control.

Figure 4.
S-500 and S-1000 mounted for vibration test in z-axis.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 1 av of
Dato Date

1201.07.2014

Appendix 1
Test Equipment

22

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

1

Side Page 2 av of
Dato Date

Vibrator
Int.no.0883
Int.no 15943
Int.no.1974
Int.no.1975
Int.no.972
Int.no.173699
Int.no.423352
Int.no.423354
Int.no.448239

Vibrator
AmplifierVibration control sys
Charge amplifier
Charge amplifier
Charge amplifier
Accelerometer
Accelerometer
Accelerometer
Accelerometer

LDS 964 LSLE 4014 Edp550
BK 2626
BK 2626
BK 2626
BK737
E628
E629
BK647

201.07.2014

Cal.: N.A.
Cal.: N.A.
Cal.: 09.05.14
Cal.: 13.12.13
Cal.: 02.10.13
Cal.: 13.12.13
Cal.: 02.04.14
Cal.: 19.02.14
Cal.: 04.06.14
Cal.: 03.10.13

22

Interval: 6 months
Interval: 12 months
Interval: 12 months
Interval: 12 months
Interval: 12 months
Interval: 12 months
Interval: 12 months
Interval: 12 months

Vibration fixtures were supplied by Nortek.
Table 1: Measurement uncertainty figures.
Parameter
Uncertainty
Acceleration+/- 2,95 %Coverfactor k2Remarks

Figures based on specification andfigures from calibration of individual
items in measurement chain.
Frequency+/- 0.01 %2Figures based on specification.
* Details from measurement uncertainty calculations can be forwarded on request.

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 1 av of
Dato Date

Appendix 2Test Records

2801.07.201438

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32Side Page 2 av of
Dato Date

801.07.2014

X-axis, S-55, S-500 and S-1000.
100.0
RMS: 2.94416
10.0LogMag, g²/Hz

1.0

100.0m

10.0m

1.0m

100.0u

10.0u
10.0

100.0

500.0

Hz

10.0K
1.0K
100.0

RMS: 4.3856

MP2LogMag, g²/Hz

10.0

Measurement2
X: 106.176
Y: 0.455808
Measurement2
X: 425.975
Y: 0.00883816
dX: 319.799
dY: 0.44697

1.0
100.0m10.0m
1.0m
100.0u
10.0u
10.0

100.0
Frequency, Hz

500.08

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32Side Page 3 av ofDato Date801.07.2014

X-axis, S-55, S-500 and S-1000.
100.0
RMS: 7.10244

10.0

LogMag, g²/Hz

MP3
1.0

Measurement3
X: 258.128
Y: 1.24047

100.0m

10.0m

1.0m
10.0

100.0

500.0

Frequency, Hz

100.0
RMS: 7.89553
10.0

LogMag, g²/Hz

1.0

MP4

100.0m

10.0m
Measurement5
X: 265.121
Y: 1.26304

1.0m

100.0u

10.0u
10.0

100.0
Frequency, Hz

500.0

8

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32Side Page 4 av ofDato Date

801.07.2014

Y-axis, S-55, S-500 and S-1000
100.0
RMS: 2.93264
10.0

LogMag, g²/Hz

1.0

100.0m

10.0m

1.0m

100.0u

10.0u
10.0

100.0

500.0

100.0

500.0

Hz

10.0K
1.0K
100.0

MP2
RMS: 6.34707

LogMag, g²/Hz

10.0
1.0
100.0m10.0m
1.0m
100.0u10.0u
10.0

Measurement2
X: 94.7316
Y: 0.308638
Measurement2
X: 119.527
Y: 1.62723
dX: 24.7955
dY: 1.31859
Frequency, Hz8

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32Side Page 5 av ofDato Date

801.07.2014Y-axis, S-55, S-500 and S-1000.
100.0
RMS: 7.16123

LogMag, g²/Hz

10.0

1.0

MP3

Measurement3
X: 250.498
Y: 1.19298

100.0m

10.0m

1.0m
10.0

100.0

500.0

Frequency, Hz

100.0
RMS: 7.20031
10.0
MP4

LogMag, g²/Hz

1.0

100.0m

10.0m

1.0m

Measurement5
X: 254.949
Y: 1.26611

100.0u

10.0u
10.0

100.0
Frequency, Hz

500.08

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32Side Page 6 av ofDato Date

801.07.2014

Z-axis, S-55.
100.0
RMS: 2.93204
10.0

LogMag, g²/Hz

1.0

100.0m

10.0m

1.0m

100.0u

10.0u
10.0

100.0

500.0

100.0

500.0Hz

10.0K
1.0K
100.0

MP2
RMS: 11.089

LogMag, g²/Hz

10.0
1.0
100.0m10.0m
1.0m
100.0u10.0u
10.0

Measurement2
X: 86.4665
Y: 0.355272
Measurement2
X: 127.157
Y: 3.81958
dX: 40.6901
dY: 3.46431
Frequency, Hz8

Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32Side Page 7 av of
Dato Date

801.07.2014

Z-axis, S-500 and S-1000.
100.0RMS: 2.93393
10.0

LogMag, g²/Hz

1.0

100.0m

10.0m

1.0m

100.0u

10.0u
10.0

100.0

500.0

100.0

500.0

Hz

100.0
RMS: 8.87468

LogMag, g²/Hz

10.0

1.0

MP3

Measurement3
X: 446.32
Y: 1.6566

100.0m

10.0m

1.0m
10.0
Frequency, Hz

8Rapportnr. Report no.

3000-14-017757
Revisjonsnr Rev.nr. Vedlegg Appendix

32

Side Page 8 av of
Dato Date

801.07.2014

Z-axis, S-500 and S-1000.
100.0
RMS: 6.9163
10.0MP4

LogMag, g²/Hz

1.0

100.0m

10.0m

1.0m

Measurement5
X: 469.208
Y: 1.02738

100.0u

10.0u
10.0

100.0
Frequency, Hz

500.08