Specification List of ERL-110C Series
STANDARD SPECIFICATIONITEMSPECIFICATIONClassRefer to [ ERL Specification List ]Support system

Pivot system

Swing angle of meter

205°

Dimension meter from front

110×110mm

Length of scale

164mm

Scale plate

WhitePointerInstrument

Lance type, Black

Highest record

Lance type, Red

Reset record

Manual rest with push button

Response time

When apply step input, final steady state value is measure up to 90% in time

Installation posture

Vertical (⊥)

Material panel

Iron plate and non-iron plate

Thickness panel

Below 10mmType of output signal

Passing type not contact output (holding type)

Pointer moving range

Passing full scale type

Setting range

Full scale

Pickup value tolerance±1.0% of scale length

Dead band

1.0% of scale length

Min. setting width

3% of scale length

Relay operation control system

H (upper limit)

Setting index

Triangle, Red

Relay control power

AC110V or AC220V (4.5VA): +10%-15% (50/60Hz)
DC100V or DC110V (30.VA): -20%+30% ( For CERL-110CH Only)

Contact structure

1C contact

Contact capacity

AC200V, 0.5A, (resistance load) ; DC30V, 2A (resistance load)
Black: (Munsell N1.5)

Color of cover

Dark blue: (Munsell 7.5BG4/1.5)

Material of case

Cover: Methacrylic acid resin molding (Antistatic prevention treatment)
Base: Flame-retarded ABS resin

Insulation resistance

Between electric circuit and outer case

DC500V, 50MΩ or more

Voltage test

Between electric circuit and outer case

AC2210V, between 5sec.

Safetyrequirement

Standard

JISC 1010

Insulation

Between electric circuit and outer case: Base of insulation

Use

For indoor use (Cubicle etc.)

High altitude2000 or less

Pollution

Pollution level 2

Measure category

CATⅢ

Max. circuit voltageOperated temperature /humidity range
Storage temperature range

300V (Ammeter)-10~55℃, Average day temperature is 40℃ or less, 25~85% RH-20~70℃

9Specification List of ERL-110C Series
STANDARD SPECIFICATION
ITEM

Scale

SPECIFICATION

Color line

Red, Green, Yellow (Please specify)

Color area (bar)Red, Green, Yellow (Please specify)

Double scale

Please specify

Double seal

Please specify

Max. scale division

110 angle: 100 division

Special scale

Please specify

Tropical specification
Rest equipment of record pointerElectromagnetic Reset

Rust preventative,「FOR TROPICS」will display at the name plate
Voltage

AC110V ±15% ; DC110V, 48V, 24V ±20% (Please specify)

Consumption VA

10VA

Apply time

Below 1 min.

Installation posture

Horizontal, inclination installation (specify the angle)

Material of flame retardant

Cover

Polycarbonate resin

Dimensions

ITEM TO SPECIFY WHEN PURCHASE
1. Type

5. Record pointer rest system

2. Max. scale value(a) Manual rest type

3. Response time(b) Manual and Electromagnetic combined usage type

4. Rating of CT or VT

6. Aux. power for manual & electromagnetic combined usage type
7. Aux. power
8. Option (Refer to special specification)

10Specification List of ERL-110C Series
LIST OF SPECIFICATION
TypePrinciple

Moving coilresistance

Voltage

Current

20mA－

Below 500Ω

50mA

－200Ω

100mA

－100Ω

0.1, 0.2

100µA~10A

－

60mV

0.2

10V~300V

20mA－

MTERL-110CH

60mV~10V

1mA－

CERL-110CH

0.1A~10A－

Below 6VA

100V~300V

Below 6VA

－2VA

1VA

1.5
Transducer

MTERL-110CH

Moving coil

MERL-110CH

DC Voltmeter

1.5

AC Ammeter
Rectifier
AC Voltmeter

Watt
Meter

Var
Meter

Power
FactorMeter

Singlephase
3phaseSinglephase
3phaseSinglephase
3phaseClassMERL-110CH

DC AmmeterTransducer

RecordPointer
Accuracy

VA or Internal

Model

1.5± 2.0%± 2.0%

Transducer

WERL110CH-33

1.5± 2.0%

110V, 5A
220V, 5A

WVERLTransducer

110CH-12
WVERL-

1.5± 2.0%

110V, 5A
220V, 5A

110CH-33
PERLTransducer

110CH-12
PERL110CH-33

5.0± 6.0%

Responsetime (sec)

0.20.2± 2.0%

CERL-110CH
WERL110CH-12

Max. Input
and Range

110V, 5A
220V, 5A

Each phase
2VA

Each phase
1VA

2VA

1VA

Each phase
2VA

Each phase
1VA

2VA

1VA

Each phase

Each phase

2VA

1VA

0.1, 0.15, 0.2

Transducer

Note－Auxiliary Power
AC100V or AC200V

Auxiliary Power
AC100V or AC200V－

0.5~300.1, 0.15, 0.2－

0.5~30
WT-83M-12
0.5~30
WT-83M-33

2~30

WVT-83M-12

Specifyfrequency

WVT-83M-33

Specifyfrequency

PT-83M-12

Specifyfrequency

PT-83M-33

Specifyfrequency

2~30

Note:
Electromagnetic reset voltage: AC110V±15%, DC110, 48, 24V ±20%
It is possible to reset the record pointer by impressed electromagnetic reset voltage.
Record pointer will be reset in 1 sec. please do not impress it more than 1 min.

Connection Diagram

Behavior of Alarm Output Contact（－）（＋）

11