RSE-0698-3
GAS MONITOR
RI-600(ppm)SPECIFICATION
Detection principle
Gas to be detected
DisplayNon-dispersive infrared absorption method
Carbon dioxide (CO2)
LCD digital display (Five-digit, seven-segment/ green, orange andred backlight)

Detection range※Display resolution
Detection method
Alarm setpoint
Indicate accuracy

0～2000ppm/0～5000ppm/0～10000ppm
0～2000ppm：1ppm/2000ppm～10000ppm：10ppm
Diffusion type
1000ppm(1st)/1000ppm(2nd)[Standard setting]Within ±5% F.S (For the gas density signal output)*The backlight is off during standard setting.

(under an identical condition)

Response time

Within 60sec(T90)(under an identical condition)

Gas alarm typeGas alarm display

Two-step alarm (H-HH)
1st: Lights up the concentration display and backlight (orange),buzzer
2nd: Lights up the concentration display and backlight (red),buzzer

Gas alarm pattern
Gas alarm contact※

Non latching (auto-reset)
No-voltage contact 1a or 1b and normally de-energized (energizedin response to an alarm)
System abnormalities/sensor connection abnormalities
Alarm detail display and backlight blinking (orange), buzzer*The backlight and buzzer are off during standard setting.

Fault alarm/self diagnosis
Fault alarm display*The backlight is off during standard setting.

Fault alarm pattern
Non latching (auto-reset)
Contact capacity
125VAC･1A or 30VDC･1A (Resistance load)
Transmission
4～20mADC (no-insulation/load resistance under 300Ω)
specifications
Power supply※
100-120VAC±10% (50/60Hz) or 24VDC±10%Power consumption
AC specification:Max.6VA/DC specification:Max.4W
Initial clear
Approx. 25 seconds
Warming-up time
Approx. 30 minutes
Operating temperatures
0～40ºC (at a constant condition)
Operating humidities
Below 90%RH (Non-condensing)
Structure
Wall mounting type
Outline dimensions
Approx. 80(W)x120(H)x 35.5(D)mm (projection portions excluded)
WeightAC specification:Approx.200g/DC specification:Approx.180g※ Please specify your request when ordering.
Outline Drawings<Main body>

RSE-0699-4
GAS MONITOR
RI-600(vol%)SPECIFICATION
Detection principle
Gas to be detected
DisplayNon-dispersive infrared absorption method
Carbon dioxide(CO2)
LCD digital display(Five-digit, seven-segment / green, orange andred backlight)*The backlight is off during standard setting.

Detection range※Display resolution
Detection method
Alarm setpoint
Indicate accuracy

0 - 2vol% / 0 - 5vol%0 - 2vol%：0.005vol% / 2 - 5vol%：0.010vol%Diffusion type0 - 2vol%：1.0vol%(1st)/1.0vol%(2nd)[Standard setting]0 - 5vol%：2.5vol%(1st)/2.5vol%(2nd)[Standard setting]Within ±5% F.S (For the gas density signal output)(under an identical condition)

Response time

Within 60sec(T90)(under an identical condition)

Gas alarm typeGas alarm display

Two-step alarm(H-HH)
1st: Lights up the concentration display and backlight(orange),buzzer
2nd: Lights up the concentration display and backlight(red),buzzer

Gas alarm pattern
Gas alarm contact※

Non latching(auto-reset)
No-voltage contact 1a or 1b and normally de-energized(energizedin response to an alarm)
System abnormalities/sensor connection abnormalities
Alarm detail display and backlight blinking(orange), buzzer*The backlight and buzzer are off during standard setting.

Fault alarm/self diagnosis
Fault alarm display*The backlight is off during standard setting.

Fault alarm patternContact capacity
Transmission specifications
Cable for gas detector head※
Power supply※Power consumption
Initial clear
Warming-up time
Operating temperatures
Operating humidities
Structure※
Outline dimensions

WeightNon latching(auto-reset)
125VAC･1A or 30VDC･1A(Resistance load)
4 - 20mADC(no-insulation/load resistance under 300Ω)
Dedicated cable for remote(3, 5, 10 or 20 m)
100 - 120VAC±10%(50/60Hz) or 24VDC±10%
AC specification:Max.6VA/DC specification:Max.4W
Approx. 25 seconds
Approx. 30 minutes
0 - 40ºC(at a constant condition)
Below 90%RH(Non-condensing)
Wall mounting type, sensor integrated type or remote type
Main unit: Approx. 80(W)x120(H)x 35.5(D)mm
Remote sensor: Approx. 40(W)x96(H)x35.5(D)mm(projection portionsexcluded)
AC specification: Approx. 200g
DC specification: Approx. 180g
Remote sensor part: Approx. 55g(cable excluded)※ Please specify your request when ordering.