K-ASSAY 

KAMIYA BIOMEDICAL COMPANY

KAMIYA BIOMEDICAL COMPANY

Human Cu/Zn SOD ELISA

For the quantitative determination of human Cu/Zn SOD in cell
culture supernatants, serum, plasma, urine, amniotic fluid, fetal umbilical veinblood and other body fluids

Cat. No. KT-034
For research use only, not for use in diagnostic procedures.

1Rev. 12609034

K-ASSAY 

KAMIYA BIOMEDICAL COMPANY

Product Information
Human Cu/Zn SOD ELISA
Cat. No. KT-034

PRODUCTThe K-ASSAY  Human Cu/Zn SOD ELISA is an enzyme-linked immunosorbent assay for the quantitative determinationof human Cu/Zn SOD in cell culture supernatants, serum, plasma, urine, amniotic fluid, fetal umbilical vein blood andother body fluids. The Human Cu/Zn SOD ELISA is for research use only. Not for use in diagnostic procedures.

DESCRIPTIONSuperoxide dismutases (SODs) are a unique family of metalloproteins that catalyze the dismutation of superoxide anion
radicals (O2 ) to oxygen (O2) and hydrogen peroxide (H2O2)--

O2 + O2 + 2H+

H 2O 2 + O 2

SOD is ubiquitous in oxygen metabolizing cells protecting these cells against direct and indirect oxygen-mediated free
radical damage. Four types of SOD have been defined on the basis of distinctions in their metal cofactors anddistribution: Manganese (Mn SOD) principally located in the matrix of mitochondria of all aerobes, copper/zinc (Cu/Zn
SOD) mainly present in the cytoplasm of eukaryotic cells, iron (Fe SOD), predominantly in the cytosol, chloroplasts or
mitochondria of prokaryotes as well as extracellular (EC SOD), which is found in the extracellular fluids or membraneassociated in mammals.
The properties of Cu/Zn superoxide dismutase are quite different from those of the manganese or iron enzymes.
Sequence analysis has indicated a homology between Mn and Fe class enzymes but these have no homology with the
Cu/Zn enzyme. The human Cu/Zn superoxide dismutase is a dimeric protein composed of 2 subunits of 153 amino acidresidues and a molecular weight of 16 kDa each. Dissociation of the subunits is facilitated by alkylation of the two
sulfhydryl groups in the protein or by removal of the copper and zinc ions. The human Cu/Zn SOD gene has been
localized to chromosome 21q22.1. Cu/Zn SOD gene expression is induced by mediators of oxidative stress like sulfhydryl
antioxidants, interleukin-1, tumor necrosis factor. Constitutive expression of copper and zinc SOD mRNA is highest in
dividing cells. Induction of Cu/Zn SOD expression resulting in elevated levels of Cu/Zn SOD in human body fluids is ofvalue for measuring the activity of different diseases.

2

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

PRINCIPLE
Coated Microwell
An anti-Cu/Zn SOD monoclonal coating antibody is adsorbed onto
microwells.

- Monoclonal Coating Antibody

First Incubation
Cu/Zn SOD present in the sample or calibrator binds to antibody
adsorbed to the microwells; a HRP-conjugated monoclonal antiCu/Zn SOD antibody is added and binds to Cu/Zn SOD captured by
the first antibody.

- Cu /Zn SOD- HRP- Conjugate

Following incubation unbound enzyme-conjugated anti-Cu/Zn SOD
is removed during a wash step and substrate solution reactive with
HRP is added to the wells.

A colored product is formed in proportion to the amount of Cu/Zn
SOD present in the sample. The reaction is terminated by additionof acid and absorbance is measured at 450 nm. A calibration curve
is prepared from seven Cu/Zn SOD calibrator dilutions and Cu/Zn
SOD sample concentration determined.

Second Incubation- Substrate- Reacted Substrate

3

Rev. 12609034

KAMIYA BIOMEDICAL COMPANYCOMPONENTS
 1 aluminum pouch with Microwell Plate (12 x 8-well strips) coated with monoclonal antibody to human Cu/Zn SOD
 2 vials (20 µL) HRP-Conjugate anti-Cu/Zn SOD monoclonal (mouse) antibody
 2 vials (0.5 mL) Cu/Zn SOD Calibrator, 5 ng/mL
 1 bottle (50 mL) Wash Buffer Concentrate 20X (PBS with 1% Tween 20)
 1 vial (5 mL) Assay Buffer Concentrate 20X (PBS with 1% Tween 20 and protein stabilizer)
 1 bottle (5 mL) Phosphate Buffered Saline Concentrate (PBS), 20X
 1 vial (15 mL) Substrate Solution
 1 vial (15 mL) Stop Solution (1M Phosphoric acid) 2 adhesive Plate Seals

MATERIALS REQUIRED BUT NOT PROVIDED
 5 mL and 10 mL graduated pipettes
 5 µL to 1,000 µL adjustable single channel micropipettes with disposable tips
 50 µL to 300 µL adjustable multi-channel micropipette with disposable tips
 Multi-channel micropipette reservoir
 Beakers, flasks, cylinders necessary for preparation of reagents
 Device for delivery of wash solution (multi-channel wash bottle or automatic wash system) Microplate reader capable of reading at 450 nm (620 nm as optional reference wavelength)
 Distilled or de-ionized water Statistical calculator with program to perform linear regression analysis.

STORAGEStore kit reagents at 4°C as indicated. Immediately after use remaining reagents should be returned to cold storage (4°C).
Expiration date of the kit and reagents is stated on labels. The expiration date of the kit components can only beguaranteed if the components are stored properly, and if, in case of repeated use of one component, the reagent is notcontaminated by the first handling.

SPECIMEN COLLECTION
Cell culture supernatants, human serum, EDTA, citrate or heparinized plasma, urine, amniotic fluid, fetal umbilical veinblood, or other body fluids will be suitable for use in the assay. Remove serum or plasma from the clot or red cells,
respectively, as soon as possible after clotting and separation.
Samples containing a visible precipitate must be clarified prior to use in the assay. Do not use grossly hemolyzed or
lipemic specimens.
Samples should be aliquoted and must be stored frozen at -20°C to avoid loss of bioactive Cu/Zn SOD. If samples are to
be run within 24 hours, they may be stored at 4°C. Avoid repeated freeze-thaw cycles. Prior to assay, frozen sample
should be brought to room temperature slowly and mixed gently.
For sample stability, see page 11.

PRECAUTIONS FOR USE-

All chemicals should be considered as potentially hazardous. We therefore recommend that this product is handled
only by those persons who have been trained in laboratory techniques and that it is used in accordance with theprinciples of good laboratory practice. Wear suitable protective clothing such as laboratory overalls, safety glasses
and gloves. Care should be taken to avoid contact with skin or eyes. In the case of contact with skin or eyes wash
immediately with water.

-

Reagents are intended for research use only and are not for use in diagnostic or therapeutic procedures.

-

Do not mix or substitute reagents with those from other lots or other sources.

4

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY-

Do not use kit reagents beyond expiration date on label.

-

Do not expose kit reagents to strong light during storage or incubation.

-

Do not pipette by mouth.

-

Do not eat or smoke in areas where kit reagents or samples are handled.

-

Avoid contact of skin or mucous membranes with kit reagents or specimens.

-

Rubber or disposable latex gloves should be worn while handling kit reagents or specimens.

-

Avoid contact of substrate solution with oxidizing agents and metal.

-

Avoid splashing or generation of aerosols.

-

In order to avoid microbial contamination or cross-contamination of reagents or specimens, which may invalidate theassay, use disposable pipette tips and/or pipettes.

-

Use clean, dedicated reagent trays for dispensing the conjugate and substrate reagent.

-

Exposure to acids will inactivate the conjugate.

-

Distilled water or de-ionized water must be used for reagent preparation.

-

Substrate solution must be at room temperature prior to use.

-

Decontaminate and dispose samples and all potentially contaminated materials as if they could contain infectious
agents. The preferred method of decontamination is autoclaving for a minimum of 1 hour at 121.5°C.

-

Liquid wastes not containing acid and neutralized waste may be mixed with sodium hypochlorite in volumes suchthat the final mixture contains 1.0% sodium hypochlorite. Allow 30 minutes for effective decontamination. Liquid
waste containing acid must be neutralized prior to the addition of sodium hypochlorite.

PROCEDURES
PREPARATION OF REAGENTS
Buffer Concentrates should be brought to room temperature and should be diluted before starting the test procedure. If
crystals have formed in the Buffer Concentrates, warm them gently until they have completely dissolved.
A. Wash Buffer (1X)
Pour entire contents (50 mL) of the Wash Buffer Concentrate (20X) into a clean 1,000 mL graduated cylinder. Bring final
volume to 1,000 mL with distilled or de-ionized water. Mix gently to avoid foaming.
Transfer to a clean wash bottle and store at 2° to 25°C. Please note that the Wash Buffer (1X) is stable for 30 days.
Wash Buffer (1X) may be prepared as needed according to the following table:
Numberof Strips

Wash Buffer Concentrate (20X)(mL)

Distilled Water
(mL)

1-6
1 - 12

25
50

475
950

B. Assay Buffer (1X)
Add contents of Assay Buffer Concentrate (20X) (5.0 mL) to 95 mL distilled or de-ionized water and mix gently to avoid
foaming. Store at 4°C. Please note that the Assay Buffer (1X) is stable for 30 days.
Assay Buffer (1X) may be prepared as needed according to the following table:
Numberof Strips

Assay Buffer Concentrate (20X)(mL)

Distilled Water
(mL)

1-6
1 - 12

2.5
5.0

47.5
95.0

C. Phosphate Buffered Saline (PBS) (1X)
Mix the contents of the bottle well. Add contents of PBS concentrate (20X) (5.0 mL) to 95 mL distilled or de-ionized waterand mix gently to avoid foaming. Store at 4°C. Please note that the PBS (1X) is stable for 30 days.
PBS (1X) may be prepared as needed according to the following table:

5

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

Numberof Strips

PBS Concentrate (20X)(mL)

Distilled Water
(mL)

1-6
1 - 12

2.5
5.0

47.5
95.0

D. Preparation of HRP-Conjugate
Dilute the HRP-Conjugate 1:5 just prior to use by adding 80 µL Assay Buffer (1X) to the tube containing the HRPConjugate concentrate. Mix the contents of the tube well. Make a further 1:100 dilution with Assay Buffer (1X) in a clean
plastic tube. Please note that the HRP-Conjugate should be used within 30 minutes after dilution. The HRP-Conjugate
may be prepared as needed according to the following table:
Numberof Strips

Prediluted (1:5) HRP-Conjugate
(mL)
0.03
0.06

1-6
1 - 12

Assay Buffer (1X)
(mL)
2.97
5.94

ASSAY PROTOCOLS
a. Predilute serum or plasma samples 1:20 with PBS (1X) according to the following dilution scheme:
10 µL Sample + 190 µL PBS (1X)
7

For fetal umbilical vein blood first adjust samples to 2 x 10 erythrocytes/mL. Then proceed as above.
b. Determine the number of Microwell Strips required to assay the desired number of samples plus appropriate number
of wells needed for running blanks and calibrators. Each sample, calibrator, blank, and optional control sample should
be assayed in duplicate. Remove extra Microwell Strips from holder and store in the foil bag sealed tightly with theprovided desiccant at 4°C.
c.

Wash the Microwell Strips twice with approximately 400 µL Wash Buffer per well with thorough aspiration of microwell
contents between washes. Allow the Wash Buffer to sit in the wells for about 10-15 seconds before aspiration.Take
care not to scratch the inner surface of the microwells.
After the last wash, empty wells and tap Microwell Strips on absorbent pad or paper towel to remove excess Wash
Buffer. Use the Microwell Strips immediately after washing or place upside down on a wet absorbent paper for not
longer than 15 minutes. Do not allow wells to dry.

d. Add 100 µL of PBS (1X), in duplicate, to the calibrator wells, leaving the first wells (5 ng/mL) empty. Prepare
calibrator dilutions by pipetting 200 µL of Cu/Zn SOD Calibrator, in duplicate, into well A1 and A2 (see Figure 1 and
2). Transfer 100 µL to wells B1 and B2 respectively. Mix the contents by repeated aspiration and ejection andtransfer 100 µL to well C1 and C2 respectively. Take care not to scratch the inner surface of the microwells.
Continue this procedure four times, creating two rows of Cu/Zn SOD Calibrator dilutions ranging from 5 to 0.08 ng/mL.
Discard 100 µL of the contents from the last microwells (G1, G2) used.
Figure 1.

Preparation of Cu/Zn SOD Calibrator dilutions:transfer 100 µL

discard
100 µL

Cu/Zn SOD
Calibrator 200 µL

A1

B1

C1

D1 - G1

100 µL PBS (1X)

6

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

Figure 2.

Diagram depicting an example of the arrangement of blanks, calibrators and samples in the Microwell
Strips:1

2

3

A

Calibrator 1(5 ng/mL)

Calibrator 1(5 ng/mL)

Sample 1

Sample 1

B

Calibrator 2(2.5 ng/mL)

Calibrator 2(2.5 ng/mL)

Sample 2

Sample 2C

Calibrator 3(1.25 ng/mL)

Calibrator 3(1.25 ng/mL)

Sample 3

Sample 3

D

Calibrator 4(0.63 ng/mL)

Calibrator 4(0.63 ng/mL)

Sample 4

Sample 4

E

Calibrator 5(0.31 ng/mL)

Calibrator 5(0.31 ng/mL)

Sample 5

Sample 5F

Calibrator 6(0.16 ng/mL)

Calibrator 6(0.16 ng/mL)

Sample 6

Sample 6G

Calibrator 7(0.08 ng/mL)
Blank

Calibrator 7(0.08 ng/mL)
Blank

Sample 7

Sample 7

Sample 8

Sample 8

H4

e.

Add 100 µL of PBS (1X), in duplicate, to the blank wells.

f.

Add 90 µL of PBS (1X) to all wells designated for samples.

g.

Add 10 µL of each prediluted Sample, in duplicate, to the designated wells.

h.

Prepare HRP-Conjugate. (Refer to PREPARATION OF REAGENTS).

i.

Add 50 µL of diluted HRP-Conjugate to all wells, including the blank wells.

j.

Cover with a Plate Seal and incubate at RT (18° to 25°C) for 1 hour, if available on a rotator set at 100 rpm.

k.

Remove Plate Seal and empty wells. Wash Microwell Strips 3 times according to point c. of the Assay Protocols.
Proceed immediately to the next step.

l.

Pipette 100 µL of TMB Substrate Solution to all wells, including the blank wells.

m.

Incubate the Microwell Strips at room temperature (18° to 25°C) for about 10 minutes. Avoid direct exposure tointense light. The color development on the plate should be monitored and the substrate reaction stopped(see point n. of this protocol) before positive wells are no longer properly recordable. Determination of theideal time period for color development has to be done individually for each assay. It is recommended to add
the stop solution when the highest calibrator has developed a dark blue color. Alternatively, the color development
can be monitored by the ELISA reader at 620 nm. The substrate reaction should be stopped as soon as an OD of
0.9 – 0.95 is reached for calibrator 1.

n.

Stop the enzyme reaction by quickly pipetting 100 µL of Stop Solution into each well, including the blank wells. It is
important that the Stop Solution is spread quickly and uniformly throughout the microwells to completely inactivate
the enzyme. Results must be read immediately after the Stop Solution is added or within one hour if the Microwell
Strips are stored at 4°C in the dark.

o.

Read absorbance of each microwell on a microplate reader using 450 nm as the primary wavelength (optionally 620
nm as the reference wavelength; 610 nm to 650 nm is acceptable). Blank the microplate reader according to the
manufacturer's instructions by using the blank wells. Determine the absorbance of both the samples and the Cu/Zn
SOD calibrators.

Note:

In case of incubation without shaking the obtained O.D. values may be lower than indicated below.
Nevertheless the results are still valid.

7

Rev. 12609034

KAMIYA BIOMEDICAL COMPANYCALCULATIONS-

Calculate the mean absorbance values for each set of duplicate calibrators and samples. Duplicates should be within
20 percent of the mean.

-

Create a calibration curve by plotting the mean absorbance for each calibrator concentration on the ordinate against
the Cu/Zn SOD concentration on the abscissa. Draw a best-fit curve through the points of the graph (a 5-parameter
curve fit is recommended).

-

To determine the concentration of circulating Cu/Zn SOD for each sample, first find the mean absorbance value on
the ordinate and extend a horizontal line to the calibration curve. At the point of intersection, extend a vertical line tothe abscissa and read the corresponding Cu/Zn SOD concentration.

-

For samples which have been diluted according to the instructions given in this package insert (1:200), theconcentration read from the calibration curve must be multiplied by the dilution factor (x 200).

-

Note:

-

It is suggested that each testing facility establishes a control sample of known Cu/Zn SOD concentration and runs
this additional control with each assay. If the values obtained are not within the expected range of this control, theassay results may be invalid.

-

A representative calibration curve is shown in Figure 3. This curve cannot be used to derive assay results. Everylaboratory must prepare a calibration curve for each assay run.

Figure 3.

Calculation of samples with a concentration exceeding calibrator 1 may result in incorrect, low
Cu/Zn SOD levels (Hook Effect). Such samples require further dilution with PBS (1X) in order toprecisely quantitate the actual Cu/Zn SOD level.

Representative calibration curve for Human Cu/Zn SOD ELISA

Cu/Zn SOD was diluted in serial two-fold steps in PBS (1X). Do not use this calibration curve to derive assay results. A
calibration curve must be run for each assay run.

8

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

Typical data using the Human Cu/Zn SOD ELISA
Measuring wavelength:
Reference wavelength:

Calibrator
1

2

3

4

5

6

7

Blank

450 nm
620 nmCu/Zn SOD
Concentration(ng/mL)O.D.
(450 nm)

O.D.
Mean

C.V.
(%)

5.00

1.942

1.886

4.2

5.00

1.829

2.50

1.006

0.981

3.6

2.50

0.956

1.25

0.568

0.542

6.8

1.25

0.516

0.63

0.309

0.299

4.7

0.63

0.289

0.31

0.160

0.158

2.2

0.31

0.155

0.16

0.091

0.091

0.8

0.16

0.090

0.08

0.058

0.059

1.2

0.08

0.059

0.00

0.022

0.024

8.3

0.00

0.026

The OD values of the calibration curve may vary according to the conditions of assay performance (e.g. operator,
pipetting technique, washing technique or temperature effects). Furthermore shelf life of the kit may affect enzymatic
activity and thus color intensity. Values measured are still valid.

LIMITATIONS-

Since exact conditions may vary from assay to assay, a calibration curve must be established for every assay run.

-

Bacterial or fungal contamination of either samples or reagents or cross-contamination between reagents may cause
erroneous results.

-

Disposable pipette tips, flasks or glassware are preferred. Reusable glassware must be washed and thoroughly
rinsed of all detergents before use.

-

Improper or insufficient washing at any stage of the procedure will result in either false positive or false negative
results. Empty wells completely before dispensing fresh Wash Buffer, fill with Wash Buffer as indicated for each wash
cycle and do not allow wells to sit uncovered or dry for extended periods.

-

The use of radioimmunotherapy has significantly increased the number of patients with human anti-mouse IgG
antibody (HAMA). HAMA may interfere with assays utilizing mouse monoclonal antibodies leading to both false
positive and false negative results. Serum samples containing antibodies to mouse immunoglobulins can still beanalyzed in such assays when mouse immunoglobulins (serum, ascitic fluid, or monoclonal antibodies of irrelevant
specificity) are added to the sample.

PERFORMANCE CHARACTERISTICS
A. Sensitivity
The limit of detection of Cu/Zn SOD defined as the analyte concentration resulting in an absorption significantly higher
than that of the dilution medium (mean plus two standard deviations) was determined to be 0.04 ng/mL (mean of 6
independent assays).

9Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

B. Reproducibility
a. Intra-assay
Reproducibility within the assay was evaluated in three independent experiments. Each assay was carried out with 6
replicates of 8 serum samples containing different concentrations of Cu/Zn SOD. Two calibration curves were run on eachplate. Data below show the mean Cu/Zn SOD concentration and the coefficient of variation for each sample. The overall
intra-assay coefficient of variation has been calculated to be 5.1%.

Positive
SampleExperiment

Cu/Zn SOD
Concentration (ng/mL)

Coefficient of
Variation (%)

1

1
2
3

110.1
99.2
93.9

4.7
6.1
7.0

2

1
2
3

194.9
185.1
179.4

2.2
1.0
3.6

3

1
2
3

129.4
123.7
124.8

4.2
6.1
3.4

4

1
2
3

48.1
45.6
38.0

1.8
10.7
8.2

5

1
2
3

149.2
145.5
150.1

1.4
4.7
1.8

6

1
2
3

64.2
53.3
58.5

7.8
7.8
7.0

7

1
2
3

133.5
122.5
121.0

7.5
9.5
2.5

8

1
2
3

42.8
40.1
42.1

5.9
4.6
2.1

b. Inter-assay
Assay-to-assay reproducibility within one laboratory was evaluated in three independent experiments. Each assay was
carried out with 6 replicates of 8 samples containing different concentrations of Cu/Zn SOD. Two calibration curves wererun on each plate. Data below show the mean Cu/Zn SOD concentration and the coefficient of variation calculated on 18
determinations of each sample. The overall inter-assay coefficient of variation has been calculated to be 5.8%.

SampleCu/Zn SOD
Concentration (ng/mL)

Coefficient of
Variation (%)

1
2
3
4
5
6
7
8

101.1
186.5
126.0
43.9
148.3
58.7
125.7
41.6

8.2
4.2
2.4
11.9
1.6
9.3
5.4
3.4

10

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

C. Recovery Studies
The spike recovery was evaluated by spiking 4 levels of human Cu/ZnSOD into 2 normal pooled serum samples.
The amount of endogenous human Cu/ZnSOD in unspiked serum was subtracted from the spike values. The recovery
ranged from 89% to 108% with an overall mean recovery of 98%.
D. Dilution Parallelism
Four serum samples with different levels of Cu/Zn SOD were assayed at four serial two-fold dilutions (1:200-1:1,600)
covering the working range of the calibration curve. In the table below the percent recovery of expected values is listed.
Recoveries ranged from 80% to 107% with an overall mean recovery of 90%.

SampleDilution

Expected
Value

Cu/Zn SOD Concentration (ng/mL)
ObservedValue% Recovery
of Exp. Value

1

1:200
1:400
1:800
1:1,600-108.2
54.1
27.1

216.3
96.1
42.7
24.0

-89
79
89

2

1:200
1:400
1:800
1:1,600

-61.9
30.9
15.5

123.8
52.8
30.1
14.2

-85
97
92

3

1:200
1:400
1:800
1:1,600

4

1:200
1:400

-73.1
36.5
18.3
-26.5
13.3
6.6

146.1
63.7
29.2
19.6
53.0
25.9
11.1
6.5

-87
80
107
-98
83
98

1:800
1:1,600

E. Sample Stability
a. Freeze-Thaw Stability
Aliquots of serum samples (unspiked or spiked) were stored frozen at -20°C and thawed five times, and Cu/Zn SOD
levels determined. There was no significant loss of Cu/Zn SOD between 0 and 5 cycles of freezing and thawing.
b. Storage Stability
Aliquots of serum samples (spiked or unspiked) were stored at -20°C, 4°C, room temperature and at 37°C, and the
Cu/Zn SOD level determined after 24, 48 and 96 hours. There was no significant loss of Cu/Zn SOD immunoreactivity
during storage at above conditions.
F. Comparison of Serum and Plasma
From 22 individuals, serums as well as EDTA, citrate and heparin plasma obtained at the same time point were
evaluated. All these blood preparations are suitable for Cu/Zn SOD determinations. It is nevertheless highly
recommended to assure the uniformity of blood preparations used in one assay.
G. Specificity
The interference of circulating factors of the immune system was evaluated by spiking these proteins at physiologically
relevant concentrations into a human Cu/ZnSOD positive serum. There was no cross reactivity detected.

REAGENT PREPARATION SUMMARYA.

Wash Buffer (1X)

Add Wash Buffer Concentrate (20X) (50 mL) to 950 mL distilled water.
Number of Strips

Wash Buffer Concentrate(mL)

Distilled Water
(mL)

1-6
1 - 12

25
50

475
950

11

Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

B.

C.

D.

Assay Buffer (1X)

PBS (1X)

HRP-Conjugate

Add Assay Buffer Concentrate (20X) (5 mL) to 95 mL distilled water.
Number of Strips

Assay Buffer Concentrate(mL)

Distilled Water
(mL)

1-6
1 - 12

2.5
5.0

47.5
95.0

Add PBS Concentrate (20X) (5 mL) to 95 mL distilled water.
Number of Strips

PBS (20X)(mL)

Distilled Water
(mL)

1-6
1 - 12

2.5
5.0

47.5
95.0

Make a 1:5 predilution of HRP-Conjugate by adding 80 µL Assay Buffer(1X) to the tube containing the HRP-Conjugate concentrate. Make a further
1:100 dilution in Assay Buffer (1X):

Number of Strips

Prediluted (1:5)
HRP-Conjugate(mL)

Assay Buffer (1X)
(mL)

1-6
1 - 12

0.03
0.06

2.97
5.94

ASSAY PROTOCOL SUMMARY-

Predilute serum or plasma samples with PBS (1X) 1:20.

-

Wash Microwell Strips twice with Wash Buffer.

-

Add 100 µL PBS (1X), in duplicate, to calibrator wells except the first wells (5 ng/mL).

-

Pipette 200 µL Cu/Zn SOD Calibrator into the first calibrator wells and create calibrator dilutions ranging from 5 to
0.08 ng/mL by transferring 100 µL from well to well; Discard 100 µL from the last wells.

-

Add 100 µL PBS (1X), in duplicate, to the blank wells.

-

Add 90 µL PBS (1X) to the sample wells.

-

Add 10 µL 1:20 prediluted Sample, in duplicate, to designated wells.

-

Prepare HRP-Conjugate.

-

Add 50 µL diluted HRP-Conjugate to all wells.

-

Cover Microwell Strips and incubate 1 hour at RT (18° to 25°C).

-

Empty and wash Microwell Strips 3 times with Wash Buffer.

-

Add 100 µL of TMB Substrate Solution to all wells including blank wells.

-

Incubate the Microwell Strips for about 10 minutes at RT (18° to 25°C).

-

Add 100 µL Stop Solution to all wells including blank wells.

-

Blank microplate reader and measure color intensity at 450 nm.

Note:

For samples which have been diluted according to the instructions given in this package insert (1:200),
the concentration read from the calibration curve must be multiplied by the dilution factor (x 200).

12Rev. 12609034

KAMIYA BIOMEDICAL COMPANY

FOR RESEARCH USE ONLYNOT FOR USE IN DIAGNOSTIC PROCEDURES

KAMIYA BIOMEDICAL COMPANY
12779 Gateway Drive, Seattle WA 98168
Tel: (206) 575-8068 Fax: (206) 575-8094
Email: LifeScience@k-assay.com
www.k-assay.com

13

Rev. 12609034