SC 6000 System Controller
Compact, flexible, out-of-the-box programmable radiometer/photometer
Flexible Design
The SC 6000 is both a highly flexible, programmableradiometer/photometer and a system controller. The
integrated microprocessor-based controller is thecommunication hub for Labsphere’s power supplies.
With a variety of interchangeable detector assemblies,
integrating spheres and components, light sources,
and power supplies, you are able to create, upgrade, or
expand a calibration system in minutes.

Out-of-the-box Sophistication
The SC 6000 operates out-of-the-box. A 64-digit backlit
LCD display allows you to view measurement results,
select calibration factors, and set display features andremote communication parameters.
Operation features include up to 100 programmableoptical radiation calibrations, manual and auto ranging,
digital or analog display, digital filtering, 5-digit display,
display zero, and display normalization display.

SC 6000 shown with Detector

Comprehensive
Labsphere’s SC 6000 System Controller is a programmable radiometer/
photometer that allows for up to 100 programmed calibrations. Labsphere willcalibrate the SC 6000 to monitor luminous and radiant flux, luminance, radiance,
intensity, illuminance and irradiance, making it an integral component of any
calibration system.
Use the SC 6000 across a range of applications such as a uniquely designed
imaging calibration instrument, a lamp measurement photometer, or a laser
power measurement system.

FEATURES:BEST FOR:

Compatibility with Si, Ge, InGaAs, and many otherphotovoltaic detectors

Controlling Calibration
Systems

Dynamic Range of 8 Decades of Sensitivity

Photometric and Radiometric
Monitoring of Camera
Calibration Systems

Stores up to 100 Photometric and Radiometric
Calibration Constants
32 Output Lines for Computer Control of System
Peripherals

Lamp Measurement

TC/IP, DHCP Computer connections

Laser Power Measurement

Photometry

Labsphere, Inc. • 231 Shaker St. • North Sutton, NH 03260 • USA • Tel: +1 (603) 927-4266 • www.labsphere.com

SC 6000

Specifications
Model and Description
SC 6000 System Control Radiometer/Photometer

Order Number
AS-02702-000

Properties and Performance
Current Dynamic Range
Display Resolution
Maximum Reading Rate
Input
Input Connector
Input Bias
Input ImpedanceCompatible Detectors
ADC Resolution
Nonlinearity
Temperature Coefficient
Operating Environment
Calibration
Calibration Factors
Power RequirementsComputer Interface
WeightDimension W x D x HCompatible Labsphere Detectors
SDA-050-U-RTA-CX
SDA-050-P-RTA-CX
IDA-050-RTA-CXIDA-EXT-050-RTA-CX
GDA-050-U-RTA-CX

8+ decades; 1pA - 20mA
5 digits
5 readings/second
Single Channel
BNC+14 VDC Selectable0 Ω Single ended
Photovoltaic (Si, Ge, InGaAs)
16 bits± 1 count± (0.02% of reading + 0.5 counts) per °C
0° to 50°C,
0% to 70% relative humidity at 35°C
Digital factors are stored in the ROM memory
2 to 100 programmable coefficients
12 VDC, 750 mADC maximum AC/DC Converter Provided 90 ~ 264 VAC, 47 ~ 63 Hz
TCP/IP, DHCP
4.1 lbs (1.86kg)
1.75 x 8.25 x 10.5 in(4.4 x 20.9 x 26.4 cm)

Silicon Detector Assembly
Photopic Detector Assembly
InGaAs Detector Assembly
Extended InGaAs Detector Assembly
Germanium Detector Assembly

AS-02522-400
AS-02522-401
AS-02522-402
AS-02522-403
AS-02522-404

Labsphere, Inc. • 231 Shaker St. • North Sutton, NH 03260 • USA • Tel: +1 (603) 927-4266 • www.labsphere.com
231031v002 042008