LFC-050-LE Light Flux Color and Luminous
Efficacy Measurement Systems

The most economical andreliable solution for photometryof light source needs!
LightFluxColor-LE Series Systems are themost affordable and reliable systems for testing
LED lighting products. Whether you are a
manufacturer of LED luminaires, street lights,
solar powered LED lanterns, LED bulbs,
or any other type of LED lighting product,
LightFluxColor Systems will meet all your testing
requirements. LightFluxColor Systems allow
luminaire manufacturers to test LED products
for photometric performance.

LightFluxColor-LE Series Systems allow

Ideal For Flux, Color & AC/DCElectrical Characterization of:Automotive Lamps

Traffic Lighting

LED Clusters

Architectural Lighting

LED Bulbs

Railway Lighting

LightFluxColor-LE Series Systems also include

a highly sensitive mini-calibrated CCD Array
Spectrometer with spectral range from
250 to 850 nm. This low noise and broad spectral
response spectrometer provide instantaneous
measurement of radiometric, photometric, andcolor characteristics of the LED sources.

users to test AC/DC characterization of lamps
at various input frequencies along with lumens
and color parameters. Electrical lamp parameterssuch as power factor can also be tested.

The fast results from the spectrometer helps
to increase the rate of product development,
decrease the time to market, and reduce
development costs.

The NIST traceable calibrated standard included
with the system allows users to perform simple
in-house system recalibration and verification
without having to ship the system to ourmanufacturing facility. The systems are available
with 0.5 m, 1 m, 1.5 m and 2 m integrating
sphere size options to accommodate LED
chips as well as larger street lights and fixtures.
The integrating sphere is coated with
Spectraflect® coating which has up to 98%
reflectance with near Lambertian properties.
The integrating sphere coating is extremely
stable, does not yellow over time, and doesn’t
need periodic recoating. The integrating spheresare designed to measure LED sources in boththe 2π and 4π geometries.

Users of the systems are also able to perform
absorption correction with standard
LightFluxColor Systems and the system
includes application specific software.
With ability to measure light source spectrum,luminous flux, electrical characteristics andcomplete color parameters with highest degreeof accuracy and traceability, LightFluxColor
Systems have the best value of all the LED
measurement systems in the market.

Light
Measurement
SoftwareLabsphere, Inc / www.labsphere.com

Why Choose LightFluxColor• Calibrations are traceable to NIST
(USA) which are accepted andrecognized globally.
• Calibrated lamp standards NVLAP
accreditation Lab Code 200951-0(ISO 17025)• Spectral flux standards (calibration
performed at each wavelength) aresupplied with each system for highest
possible accuracy.
• Single software controls all electronics andprovides optical and electrical data.
• Competitive systems only provide
luminous flux standards with CCT
calibration which limits overallsystem accuracy.
• AC/DC operation in one packaged
testing system• An auxiliary lamp is provided forabsorption correction which is applied ateach wavelength. This improves overallmeasurement accuracy as compared toother systems on the market.
• The integrating sphere is coated with
Labsphere’s Spectraflect® that has up
to 98% reflectance and is the highest
Lambertian coating in the market.
• The sphere coating doesn’t yellow overtime and doesn’t degrade in due course.
• The integrating spheres are capable ofmeasuring in 2π and 4π geometries.
• Local support and training.

Key System Features• AC/DC operation

Detailed Technical Specifications
LFC-050 0.5 meter System includes:

LFC-050-LES
Part # AA-01166-050
Light Measurement Sphere, 0.5 meter√
Spectrally Calibrated Lamp, SCL-650√			
Auxiliary Lamp, AUX-50		
√			
Calibration and Aux Lamp Socket Assembly√			Programmable DC Power Supply, M8811, 30V, 5A√			
CCD Array Spectrometer (250 - 850 nm)		
√			Light Measurement Software		√			
System Manual and Electrical Rack√			AC Power Supply, Parwa APS6001L 0-300V,1KW√
AC Power Meter, TET P62201		
√
AC Power Supply, Chroma61603,0-300V, 1.5KW							Power Meter, Yokogawa® WT210		
Relay Controller, Arroyo RC1 		
√			
Also Available
LFC-200/050-LEX (2 m and 0.5 m sphere systems with LEX rack)			
LFC-200/050-LES (2 m and 0.5 m sphere systems with LES rack)		

*This configuration includes a 2 m and 0.5 m sphere with shared electronic rack and suitable cables.

LFC-050-LEX
Part # AA-01165-050
√
√
√
√
√
√
√
√
√
√
√
Part Number
AA-01165-000
AA-01166-000

LFC-200/050 LES LFC-200/050 LEX
System Properties and Specifications
SphereSphere Coating ReflectancePhotometric Range(Illuminant A)
Spectral Range (Spectrometer)
2π Port Size
Sphere & Crate Weight
Crate Dimension (W x D x H)
System input power requirements: 220V-240V, 16A
Spectrometer Detector
Spectral Range
Integration Time
Wavelength Accuracy
Optical Input Optical Fiber
Lamp Standard
Power
Approximate Luminous Flux
Calibration
Traceability
Power Supply (DC)Power Requirements
Current Stability
Current Rise Time
Dimension (W x D x H)
Compliance
Aux Lamp

20 in (50 cm)
98%
0.5 - 1800 lm
250* - 850 nm
6 in (15.24 cm)
60 kg
0.7 M x 1.45 M x 1.0 M

20 in (50 cm)
98%
0.5 - 1800 lm
250* - 850 nm
6 in (15.24 cm)
60 kg
0.7 M x 1.45 M x 1.0 M

Sony ILX511 linear
silicon CCD array
250* - 850 nm

Sony ILX511 linear
silicon CCD array350 - 1000 nm

1 ms – 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

1 ms - 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable
M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE
AUX-50 (50W)

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable
M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE
AUX-50 (50W)*system calibration range 350 - 1050 nm

System Optional Components

SCL-650 Cal Lamp
Replacement Aux-50 bulb
CSFS-650

Labsphere, Inc / www.labsphere.com

Part Number
AS-01335-100
LEW-00014-000
AS-01336-100• NIST traceable calibrated standards
for in-house recalibration NVLAP
accreditation Lab Code 200951-0(ISO 17025)• Measurement of electrical parameters
including power factor• Measure absolute spectrumin milliseconds• Comprehensive Light Measurement
Software capable of measuring:- Total Spectral Flux (Watts/nm)- Luminous Flux (Lumens)- Luminous Efficacy (Lumens/Watt)- Radiant Flux (Watts)- Chromaticity (x, y, u, v)
- CCT
- CRI
- Peak Wavelength- Dominant Wavelength• Spectraflect® interior coating
for sphere• Absorption correction
capabilities included

Key System Features• AC/DC operation

Detailed Technical Specifications
LFC-100 1 meter System includes:

LFC-100-LES
Part # AA-01166-100
Light Measurement Sphere, 1 meter√
Spectrally Calibrated Lamp, SCL-650√			
Auxiliary Lamp, AUX-50		
√			
Calibration and Aux Lamp Socket Assembly√			Programmable DC Power Supply, M8811, 30V, 5A√			
CCD Array Spectrometer (250 - 850 nm)		
√			Light Measurement Software		√			
System Manual and Electrical Rack√			AC Power Supply, Parwa APS6001L 0-300V,1KW√
AC Power Meter, TET P62201		
√
AC Power Supply, Chroma61603,0-300V, 1.5KW							Power Meter, Yokogawa® WT210		
Relay Controller, Arroyo RC1 		
√			
Also Available
LFC-200/050-LEX (2 m and 0.5 m sphere systems with LEX rack)			
LFC-200/050-LES (2 m and 0.5 m sphere systems with LES rack)		

*This configuration includes a 2 m and 0.5 m sphere with shared electronic rack and suitable cables.

LFC-100-LEX
Part # AA-01165-100
√
√
√
√
√
√
√
√
√
√
√
Part Number
AA-01165-000
AA-01166-000

System Properties and Specifications
SphereSphere Coating ReflectancePhotometric Range (Illuminant A)
Spectral Range (Spectrometer)
2π Port Size
Sphere Weight
System input power requirements: 220V-240V, 16A

40 in (100 cm)
98%
5 - 15,000 lm
250* - 850 nm
12 in (30 cm)
110 kg40 in (100 cm)
98%
5 - 15,000 lm
250* - 850 nm
12 in (30 cm)
110 kgSpectrometer Detector

Sony ILX511 linear
silicon CCD array
250* - 850 nm
1 ms – 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

Sony ILX511 linear
silicon CCD array350 - 1000 nm
1 ms - 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable

Compliance

M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE

M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE

Aux Lamp

AUX-50 (50W)

AUX-50 (50W)

Spectral Range
Integration Time
Wavelength Accuracy
Optical Input Optical Fiber
Lamp Standard
Power
Approximate Luminous Flux
Calibration
Traceability
Power Supply (DC)Power Requirements
Current Stability
Current Rise Time
Dimension (W x D x H)

*system calibration range 350 - 1050 nm

System Optional Components

SCL 650 cal lamp
Replacement Aux-50 bulb
CSFS-650

Labsphere, Inc / www.labsphere.com

Part Number
AS-01335-100
LEW-00014-000
AS-01336-100• NIST traceable calibrated standards
for in-house recalibration NVLAP
accreditation Lab Code 200951-0(ISO 17025)• Measurement of electrical parameters
including power factor• Measure absolute spectrumin milliseconds• Comprehensive Light Measurement
Software capable of measuring:- Total Spectral Flux (Watts/nm)- Luminous Flux (Lumens)- Luminous Efficacy (Lumens/Watt)- Radiant Flux (Watts)- Chromaticity (x, y, u, v)
- CCT
- CRI
- Peak Wavelength- Dominant Wavelength• Spectraflect® interior coating
for sphere• Absorption correction
capabilities included

Key System Features• AC/DC operation

Detailed Technical Specifications
LFC-150 1.5 meter System includes:

LFC-150-LES
Part # AA-01166-150
Light Measurement Sphere, 1.5 meter√
Spectrally Calibrated Lamp, SCL-650√			
Auxiliary Lamp, AUX-50		
√			
Calibration and Aux Lamp Socket Assembly√			Programmable DC Power Supply, M8811, 30V, 5A√			
CCD Array Spectrometer (250 - 850 nm)		
√			Light Measurement Software		√			
System Manual and Electrical Rack√			AC Power Supply, Parwa APS6001L 0-300V,1KW√
AC Power Meter, TET P62201		
√
AC Power Supply, Chroma61603,0-300V, 1.5KW							Power Meter, Yokogawa® WT210		
Relay Controller, Arroyo RC1 		
√			
Also Available
LFC-200/050-LEX (2 m and 0.5 m sphere systems with LEX rack)			
LFC-200/050-LES (2 m and 0.5 m sphere systems with LES rack)		

*This configuration includes a 2 m and 0.5 m sphere with shared electronic rack and suitable cables.

LFC-150-LEX
Part # AA-01165-150
√
√
√
√
√
√
√
√
√
√
√
Part Number
AA-01165-000
AA-01166-000

System Properties and Specifications
SphereSphere Coating ReflectancePhotometric Range (Illuminant A)
Spectral Range (Spectrometer)
2π Port Size
Sphere Weight
System input power requirements: 220V-240V, 16A

60 in (1.5 m)
98%
10 - 40,000 lm
250* - 850 nm
18 in (45 cm)
200 kg60 in (1.5 m)
98%
10 - 40,000 lm
250* - 850 nm
18 in (45 cm)
200 kgSpectrometer Detector

Sony ILX511 linear
silicon CCD array
250* - 850 nm
1 ms – 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

Sony ILX511 linear
silicon CCD array350 - 1000 nm
1 ms - 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable

Compliance

M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE

M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE

Aux Lamp

AUX-50 (50W)

AUX-50 (50W)

Spectral Range
Integration Time
Wavelength Accuracy
Optical Input Optical Fiber
Lamp Standard
Power
Approximate Luminous Flux
Calibration
Traceability
Power Supply (DC)Power Requirements
Current Stability
Current Rise Time
Dimension (W x D x H)

*system calibration range 350 - 1050 nm

System Optional Components

SCL-650 Cal Lamp
Replacement Aux-50 bulb
CSFS-650

Labsphere, Inc / www.labsphere.com

Part Number
AS-01335-100
LEW-00014-000
AS-01336-100• NIST traceable calibrated standards
for in-house recalibration NVLAP
accreditation Lab Code 200951-0(ISO 17025)• Measurement of electrical parameters
including power factor• Measure absolute spectrumin milliseconds• Comprehensive Light Measurement
Software capable of measuring:- Total Spectral Flux (Watts/nm)- Luminous Flux (Lumens)- Luminous Efficacy (Lumens/Watt)- Radiant Flux (Watts)- Chromaticity (x, y, u, v)
- CCT
- CRI
- Peak Wavelength- Dominant Wavelength• Spectraflect® interior coating
for sphere• Absorption correction
capabilities included

Key System Features• AC/DC operation

Detailed Technical Specifications
LFC-200 2 meter System includes:

LFC-200-LES
Part # AA-01166-200
Light Measurement Sphere, 2 meter√
Spectrally Calibrated Lamp, SCL-650√			
Auxiliary Lamp, AUX-50		
√			
Calibration and Aux Lamp Socket Assembly√			Programmable DC Power Supply, M8811, 30V, 5A√			
CCD Array Spectrometer (250 - 850 nm)		
√			Light Measurement Software		√			
System Manual and Electrical Rack√			AC Power Supply, Parwa APS6001L 0-300V,1KW√
AC Power Meter, TET P62201		
√
AC Power Supply, Chroma61603,0-300V, 1.5KW							Power Meter, Yokogawa® WT210		
Relay Controller, Arroyo RC1 		
√			
Also Available
LFC-200/050-LEX (2 m and 0.5 m sphere systems with LEX rack)			
LFC-200/050-LES (2 m and 0.5 m sphere systems with LES rack)		

*This configuration includes a 2 m and 0.5 m sphere with shared electronic rack and suitable cables.

LFC-200-LEX
Part # AA-01165-200
√
√
√
√
√
√
√
√
√
√
√
Part Number
AA-01165-000
AA-01166-000

System Properties and Specifications
SphereSphere Coating ReflectancePhotometric Range (Illuminant A)
Spectral Range (Spectrometer)
2π Port Size
Sphere & Crate Weight
Create Dimensions (W x D x H)
System input power requirements: 220V-240V, 16A

78 in (200 cm)
98%
15 - 50,000 lm
250* - 850 nm
20 in (50.8 cm)
500 kg
2.3 M x 2.45 M x 2.53 M

78 in (200 cm)
98%
15 - 50,000 lm
250* - 850 nm
20 in (50.8 cm)
500 kg
2.3 M x 2.45 M x 2.53 M

Spectrometer Detector

Sony ILX511 linear
silicon CCD array
250* - 850 nm
1 ms – 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

Sony ILX511 linear
silicon CCD array350 - 1000 nm
1 ms - 5 s<+/- 0.5 nm
600 um, 3 m long,(SMA Connection)

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable

35 W
650 lmSpectral Flux (W/nm)
350 - 1050 nm
NIST traceable

Compliance

M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE

M8811, DC 30V, 5A
110/220 VAC, 50/60 Hz
0.1%
35 s
8.3 x 10.5 x 3.5 in(21.1 x 26.7 x 8.9 cm)
CE

Aux Lamp

AUX-50 (50W)

AUX-50 (50W)

Spectral Range
Integration Time
Wavelength Accuracy
Optical Input Optical Fiber
Lamp Standard
Power
Approximate Luminous Flux
Calibration
Traceability
Power Supply (DC)Power Requirements
Current Stability
Current Rise Time
Dimension (W x D x H)

*system calibration range 350 - 1050 nm

System Optional Components

SCL-650 Cal Lamp
Replacement AUX-50 Bulb
CSFS-650

Labsphere, Inc / www.labsphere.com

Part Number
AS-01335-100
LEW-00014-000
AS-01336-100• NIST traceable calibrated standards
for in-house recalibration NVLAP
accreditation Lab Code 200951-0(ISO 17025)• Measurement of electrical parameters
including power factor• Measure absolute spectrumin milliseconds• Comprehensive Light Measurement
Software capable of measuring:- Total Spectral Flux (Watts/nm)- Luminous Flux (Lumens)- Luminous Efficacy (Lumens/Watt)- Radiant Flux (Watts)- Chromaticity (x, y, u, v)
- CCT
- CRI
- Peak Wavelength- Dominant Wavelength• Spectraflect® interior coating
for sphere• Absorption correction
capabilities included