Reference Guide
Grifols Fleboflex® IV Solutions
0.9% Sodium Chloride Injection, USP,in Fleboflex® plastic container

Instructions for Use
1. Overwrap Removal

2. Container Inspection> Visually inspect the container for particulate matter and discoloration.
> Check for minute leaks by squeezing inner container ﬁrmly.
> If leaks are found, discard solution as sterility may be impaired.
> If the outlet port protector is damaged, detached, or not present, discard
container as solution path sterility may be compromised.
Note: Do not administer unless the solution is clear and seal is intact.

> The overwrap serves as a moisturebarrier. It is intended to limit
evaporative moisture loss from theprimary solution container.
> The overwrap is designed to beopened by pulling apart the two
sheets of the overwrap.

> The recommended method to remove
the overwrap is to peel off sheet by
the corner while holding the othersheet at one end and carefully
remove the solution container.

Note: The overwrap should not be
removed until product is to be used.

3. To Add Medication> Prepare medication site.
> Using syringe with 19 to 22 gauge

needle, securely hold the container andinsert the needle through the center ofthe medication port and inject.

4. Attach Administration Set> Insert the needle perpendicular tothe point of addition.

> Suspend the container from eyelet
support and remove protector
from outlet port at the bottom of
the container.

> Hold the container properly andattach the administration set.

Note: The compatibility of the additives
with Sodium Chloride 0.9% must be
checked before adding a medication.

Attach Administration Set (Continued)

5. Use Non-vented Administration Sets

> While advancing through the tube, a> The Fleboﬂex container has beenslight resistance will gradually appear.
Note: To ensure proper connection
and prevent leakage, push theadministration set using a rotary
motion up to the shoulder of the spike.

> Correctly attached.

designed to work without venting.
No air is required to withdraw thesolution. Air entering through theadministration set could slow or evenstop the infusion due to air bubbles.

> Administration sets should comewithout air venting. If vented
administration sets are used, besure air venting inlet remains closed
during the infusion.

Note: After opening the container, the contents should be used immediately
and should not be stored for a subsequent infusion. Do not reconnect any
partially used containers. Discard any unused portion.

Note: Do not connect ﬂexible plastic containers in series in order to avoid air embolism due to possible residual air contained in the primary container. Use of a vented intravenous administration set withthe vent in the open position could result in air embolism. Vented intravenous administration sets with the vent in the open position should not be used with ﬂexible plastic containers.

Manufactured for ICU Medical by:
Laboratorios Grifols S.A. Calle Marte nº4
Polígono Industrial Los Llanos
30565 Las Torres de Cotillas, Murcia
SPAIN
www.grifols.com© 2023 Grifols All rights reserved US-FBX-2200013
For safe and proper use, refer to the Instructions for Use.

Distributed by:

800.824.7890 |

www.icumed.com© 2023 ICU Medical Inc. | P23-4967