RF POWER AMPLIFIER
R&K-A080M102-6060R■ Class A Linear Solid-State Amplifier■ Broadband Frequency:80MHz～1000MHz■ Output Power(CW):1kW(typ.) @1dB Comp.
■ Building-block Construction Design
APPLICATIONEMC(Electromagnetic Compatibility)
Accelerator Application
Telecommunication Test and Measurement Application
Etc...

SPECIFICATIONS ＠＋25℃

OUTLINE DRAWING※IN MILLIMETERSFRONT VIEW

SIDE VIEW

560

1000

REAR VIEWREF MON

FWD MON-60dB Coupling-60dB Coupling

REF MON IN

FWD MON IN

J001J002

ADDRESS

RF IN

GP-IB CONTROL

A080M102-6060R

CONTROL-A

RF POWER AMPLIFIERCONTROL UNIT

1 1
LOCAL

J021

REMOTERF ON/OFFST-BY

RF SW
FORWARD
REVERSE

P01
ON1kW
100W

50Ω
RF OUT

DC+12V OUTEMERGENCY IN

AC OUT-3

J003

J014
LINE INPUT
AC200V，3Φ，50/60Hz

AC OUT-2
AC OUT-1

Page UpCONTROL-B

J022

RESET

POWER

Page Down

J013
J011

J012

Debug

A080M102-6060R
RF POWER AMPLIFIER
DRIVER AMP. UNITCONTROL-A

RF IN
J121
CONTROL-B
J101
RF OUT-1

RF OUT-2

J122
AC IN

POWER
J102

A080M102-6060R

ADDRESSRF POWER AMPLIFIERFINAL AMP. UNIT

RF OUT-1

RF OUT-2RF IN

1J103

J111

CONTROL-A

AC200V IN

2Debug

CONTROL-B

ADDRESSCONTROL-A

A080M102-6060R
RF POWER AMPLIFIERFINAL COMBINER UNIT

RF IN-1

RF IN-2

FWD MONRF OUT

J202

J201
RF IN-3

J205

RF IN-4

REF MON

DC+12V IN

50Ω
J204

J203

J206

A080M102-6060R
RF POWER AMPLIFIERFINAL AMP. UNIT

RF OUT-1

RF OUT-2

RF IN

1J211

AC200V IN

2Debug

CONTROL-B

POWER

LINE INPUT
AC200V，3Φ 50/60Hz

BBW330TYPICAL PERFORMANCE (Temp @+25℃)
75

Gain Flatness

3.02.8

70

2.655
50

582.2562.0
1.8

541.440

1.2100 200 300 400 500 600 700 800 900 1000

100 200 300 400 500 600 700 800 900 1000

Frequency (MHz)Frequency (MHz)Output Power 1dB Gain Comp.

-20

Output Harmonics
Output Power = 850W

Output Harmonics (dBc)62

Output Power (dBm)

FWD LIMIT

60

Input VSWR
Output VSWR

1.6

45

63

Output Power Linearity

2.4

60

VSWR

Gain (dB)65

Input / Output VSWR

61
60
59
58
57

Output Power (dBm)

1338
1282

POWER

: 80MHz ～ 1000MHz
Frequency Range: ＋60.0dB (min.)
Small Signal Gain: ±3.0dB (max.)
Gain Flatness
Output Power (CW)
80MHz～ 90MHz : 1kW (min.) @3dB Comp.
80MHz～ 90MHz : 850W (min.) @1dB Comp.
90MHz～1000MHz : 1kW (min.) @1dB Comp.
: Class A
Operation Mode
: ＋70.0dBm (typ.)Output I.C.P.
Harmonics @90MHz～1000MHz : －20.0dBc (max.) @Po=1kW: －70.0dBc (max.) @Po=1kW
Spurious: 50Ω
Impedance: 2.0 (max.)
Input VSWR
: 2.5 (typ.) Δ1Output VSWR
: ＋3.0dBm
Maximum RF Input Power: AC200V±15％／3φ, 50／60Hz, 15kVA
AC Supply Input:0℃ to ＋35℃
Operating Temperature: －15℃ to ＋55℃
Storage Temperature: RF-IN
N -FEMALE @REAR PANEL
Connectors
RF-OUT 7/16-FEMALE @REAR PANEL: (W)560mm×(D)1000mm×(H)1282mm
Size(Excluding Projection): 305.0kg (typ.) Δ1Weight: Forced Air CoolingCooling: Over Temperature Protection
Protection Circuits
Power Supply Voltage Protection
Output Over Power Protection
Output Over Reflected Power Protection: R&K Multi Monitoring System
Other Functions
GP-IB(IEEE-488)
Safety Interlock(EMERGENCY)

52504846-30
44-40

42Frequency
80MHz200MHz
400MHz600MHz
800MHz1000MHz40-5056100 200 300 400 500 600 700 800 900 1000

Frequency (MHz)100 200 300 400 500 600 700 800 900 1000

Frequency (MHz)-20-18-16-14-12-10-8

Input Power (dBm)-6-4-20* CHANGE Δ1 7/May/2018