Static Control

Footwear Tester

Packing list

Unit, Test plate, Installation base,Knob (qty 2), Wood screw (qty 2),
Pan head screw (qty 2), AC adapter*,USB cable, Instruction manual
adapter may not be included* ACdepending on the specifications.

Specifications
RatingMeasurement voltage
Upper limit
Measurement
range

Accuracy

Lower limit
Conductivity evaluation
R < 0.1 MΩ
0.1 MΩ ≤ R ≤ 100 MΩ
100 MΩ < R

Operating environment
Environmental conditions
Dimensions
WeightOptions
DC 24 V 50 mA
DC 20 V
1000 MΩ (1 × 109 Ω)
100 MΩ (1 × 108 Ω)
10 MΩ (1 × 107 Ω)
1 MΩ (1 × 106 Ω)
0.1 MΩ (1 × 105 Ω)
R < 0.1 MΩ (1 × 105 Ω)
R < 0.1 MΩ (1 × 105 Ω)
±5%
±10%
±8%
Ambient temperature/Humidity range:
0 to 40°C, max.80% RH(without condensation)
Applicable rated pollution degree 2(according to IEC/UL61010-1)
120 (W) × 30 (H) × 185 (D) mm
0.55 kg (including installation base)Part No.
C5032
B5263
B5264

Part NameStand
Unit integration baseConnecting cable

Test plate
Cord length
Dimensions (w/o cord)
Weight (w/o cord)

1.6 m310 (W) × 18 (H) × 310 (D) mm
1.9 kgFootwear Tester
to Improve Traceability
and Work as a Combo Tester

AC adapter
Output voltage

DC 24 V

CA00551EaYa002 2024.07

Saves measurement data automatically
Meets the requirement of JIS T 8103:2010

Saves measurement data automatically and works as a combo tester
Pass/Fail indicator

Pass range indicator
Level indicator (left foot)User-friendly software

ID Management

The dedicated software makes it easy to checkmeasurement status of FG-465. The software
enables digital management of all the related
measurement information including operator IDs.

Operator IDs can be recorded as apart of measurement information. IDs
can be input by entering numericvalues, reading barcode or IC card.

*The registrations on our website are necessary to download thesoftware.

Level indicator (right foot)Pad

FG-465
Dual Foot Plate

Simple yet high visibility design

JIS T 8103:2010 requires measurement on each foot.
FG-465 measures anti-static performance of footwear on
both feet but individually, and shows the results for each foot.
The divider on the test plate prevents faulty measurement.

Separate displays of "Evaluation result", "Pass range
indicator" and "Level indicator" make it possible to checknecessary information at a glance.

Adjustable evaluation range

Installations

Evaluation range is adjustableaccording to required conditions.
The settings can be made by theDIP switches on the back side of
the unit.

The unit can be installed on adesk and wall. Also it can beattached to an optional stand.

Improved traceability
The measurement information can be automatically saved as a digital
data in a specified folder. The information consists of measurement
date, time, operator ID, and result of each foot to improve traceability.

Evaluation external output
Evaluation results can be output
through the exter nal output
terminal to interlock with otherdevices, to open a door, turn ona signal, etc.
+ Terminal- Terminal

FG465_20200630.csv

FG-465 can work as a combo
tester with FG-470, a wrist strap
tester, that makes it possible toconduct measurement, evaluation
and recording of both footwear andwrist strap at once.

An option part to integrate
FG-465 and FG-470 is available.

OptionOption123

DIP switch No. 6
OFF: External output on PASS
ON: External output on FAIL

Left side view

FG465_20200629.csv

FG465_20200701.csv

Works with wrist strap tester FG-470

image

Photocoupler output
Maximum absolute rating: DC 80 V
Maximum output current: DC 5 mA
Output time: ON for 1 second after evaluation

Saves daily measurement
results automatically(in CSV format)

FG-470C5032 Stand

B5263 Unit integration base(Connecting cable included)