Genius 3020

Nitrogen Generator for LC-MS
Part Number : 10-6320

Service Kit : 08-4782

Your local gas generation partner

Description
Combining the capacity of two separate Genius nitrogen generator
systems into one fully integrated system in the same compact
footprint, the Genius 3020 is a convenient and safe on-demandgas solution. With dual output providing independently controlled
gas sources, two LC-MS instruments can be supplied from one
Genius 3020 with ease.
For laboratories; without an external air supply and with gas
requirements above those of Genius 10 series generators, theGenius 30 series deliver nitrogen at increased ﬂow rates catering
for these specifications.
Peak Scientific’s Genius series of generators, designed mainly for
LC-MS applications, are self-contained systems featuring integrated
compressors, delivering laboratory-grade nitrogen at variouspressures and ﬂow rates required by numerous instruments.

Applications)

Nitrogen supply for up to two LC-MS instruments

Key Features
850.00 mm

600.00 mm

1303.00 mm

1303.00 mm
850.00 mm

600.00 mm

Genius 3020 Dimensions

Copyright 2015 Peak Scientific - Rev - Genius 3020 - Rev 3 - 15/06/16)

Combines two genius units into one with the same qualityoutput)

Each gas supply is controlled manually and is independent of theother)

Higher ﬂow rate 30 series generator designed for hungrier LC-MSapplications)

Compressor based solution, no need for an external air supply)

Minimal set-up required)

Highly economical source of nitrogen gas with low lifetime
running costs)

The latest generation of compressors located in an insulatedchamber reducing noise and vibration)

Service indication to allow planning of preventative maintenance)

Gas is supplied on demand so generator works to your schedule)

Wheels make it easy to manoeuvre generator around the lab

Technical Specifications

Genius 3020

Max Flow Rate2 x 32 L/min / 1.13 CFM

Max Pressure

100 psi / 6.9 bar

Max Relative Humidity

80% Non-Condensing

Max Altitude2000 Metres

Particles< 0.01μm

Gas Outlets

2x 1/4” BSPP

Phthalates

None

Suspended Liquids

None

Operating Temperature

5ºC - 35ºC / 41ºF - 95ºF

Electrical Requirements

230v 50/60Hz 12A

Power Consumption

2760 watts

Generator Dimensions (HxWxD)
Generator Weight1303 x 600 x 850 mm/ 51.3 x 23.6 x 33.4 in
189 Kg (417lbs)

Ordering Information
Part Number

10-6320

Annual Service

08-4782

Standard Maintenance Plan

09-3110Complete Maintenance Plan

09-3010

Step-Up Transformer For Use With
208V Electrical Supply

06-3210Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2015 Peak Scientific - Rev - Genius 3020 - Rev 3 - 15/06/16