Solaris 10

Nitrogen generator for ELSD andPart Number : 99-9999
Service Kit : 99-9999
CompactMass SpectrometerYour local gas generation partner

Description
Peak Scientific’s Solaris 10 nitrogen generator has been engineeredusing membrane technology to meet the gas delivery requirements
for labs using ELSD with their High Performance Liquid
Chromatography, Compact Mass Spectrometer or for TLD reader
instruments. Solaris 10 can operate with varying flow rate (up to 10L/
min of high purity nitrogen), purity (up to 99.5%) and outlet pressure
is adjustable to 100psi max offering a flexible solution for labs.
With a compact and space-saving chassis, Solaris 10 can be placedon a benchtop and paired with an additional air compressor unit
providing air to the system for labs without an in-house air supply or
who wish to contain their gas supply in a single system.

Applications)

Up to 2 x ELSD instruments)

1 x Compact Mass Spectrometer)

TLD Readers

Key Features

156 mm540 mm)

Nitrogen up to 10 L/min, up to 100 psi and up to99.5% purity)

Variable outlet flow, pressure and purity)

Compact, stackable system maximising the use ofvaluable laboratory floor or bench space)

Optional Solaris Air Compressor stackable module)

Colour indicative LED lighting for easy statusawareness)

Manufactured and tested to highest spec in UK, CE/FCC certified)

Safer and more convenient than pressurized cylinders,
dewars or bulk storage)

12 month on-site comprehensive warranty417 mm)

Solaris 10 Dimensions

Copyright © Peak Scientific Instruments Ltd 2021 — Solaris 10 Data sheet Rev 8 RSID 1605 EN —14/09/2021

Backed by [Peak Protected] global on premise
servicing and technicalTechnical Specifications

Solaris 10Max Flow Rateup to 10 l/min (input dependent)

Max Pressure

up to 100 psi / 6.9 bar (user adjustable)

Purityup to 99.5%*

Gas Outlets

1 x 1/4 " BSPP

Min/Max Air Inlet Pressure

80 psi/ 5.5 bar - 130 psi/ 8.9 bar

Min Air Inlet Flow

30 l/minMin Inlet Air Quality

ISO8573 - 1:2010 Class 1.4.1

Contaminants

Phthalate & BHT Free

Suspended Liquids

None

Operating Temperature

5°C - 35°C / 41°F - 95 °F

Electrical Requirements

Universal 100-230 VAC / 50/60 Hz / 0.03A

Power Consumption

6.9 Watts

Heat Output

25 BTU/HrGenerator Dimensions H x W x D

156 x 417 x 540 mm / 6.1 x 16.4 x 21.3”

Generator Weight

12 kg / 26.4 lbs*depending on inlet air pressure and outlet flow

Ordering Information
Part Number

65-0001

Annual Service

08-9013

Standard Maintenance Plan

09-3010Complete Maintenance Plan

09-3010

Solaris Air Compressor
Part Number

110v

230v

Annual Service Kit

65-000465-0003

08-9454

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management Systemconforms to: ISO:9001:2008

For a full list of our worldwideoffice locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com
Copyright © Peak Scientific Instruments Ltd 2021 — Solaris 10 Data sheet Rev 8 RSID 1605 EN —14/09/2021