MEASURE
MEASUREanylight source
any light source

Flashlight Performance
Performance
Test SystemSystem
Turnkeysolution toto meet
meet industry
industry standards
standards
Turnkey solution

Designed forfor thethe portableportable light industry
Designed
Designed
specifically
Designed specifically for this application, thesystem evaluates
evaluates lamp
lamp efficiency
efficiency forsystemfor development
development
andmanufactureofLED,
tungsten,xenon
and manufacture of LED, tungsten, xenon andandkrypton portableportable lighting
lighting assemblies.
assemblies.
kryptonFastFastandandaccurate
accuratetesttestresults
results
Facilitate
accurate
spectroradiometric,Facilitate accurate spectroradiometric,photometricphotometricandcolorimetriccharacterization
meetingand colorimetric characterization meetingANSI/
ANSI/
NEMA
FL1
standard
requirements
forruntime
NEMA FL1 standard requirements for run timeandandlight
lightoutput
outputmeasurements.
measurements.

Easy toto useuse
Easy
Turnkey systemsystem includes
includes Labsphere’s
Labsphere’s fast
Turnkeyfast andandaccurate
mini
spectrometer,
flashlightmeasurement
accurate mini spectrometer, flashlight measurement
integrating spheresphere spectrometer,
spectrometer, lamp
lamp standard
integrating
standard
spectral flux,
flux, absorption
absorption correction
correction lamp
spectrallamp andand switch
switchboxfordirectingelectricalcurrent
toeither
thebox for directing electrical current to either the lamp
lamp
standard or auxiliary absorption correction using thestandard or auxiliary absorption correction using thesame lamp power supply.
same lamp power supply.

MeasureMeasure(using(usingincluded
includedFPTFPTsoftware)
software)TotalLuminous
Flux(lumens)
Total Luminous Flux (lumens)Total
TotalForward
ForwardSpectral
SpectralFlux
Flux(watts/nm)(watts/nm)
Total
Radiant
Flux(watts)
Total Radiant Flux (watts)
Color
Color(CCT)
(CCT)
Lamp
Performance
Lamp Performancevs
vsTimeTime
Peak and Dominant Wavelength
Peak and Dominant Wavelength
Spectral Purity
Spectral Purity
Color Rendering Index (CR I)
Color Rendering Index (CR I)
Chromaticity Coordinates
Chromaticity Coordinates
Half-Bandwidth, Run TimeHalf-Bandwidth, Run Time

Advancing the Technology of Light: Measure. Create. Reflect.
Advancing the Technology of Light: Measure. Create. Reflect.

sales@labsphere.com
sales@labsphere.com

www.labsphere.com
www.labsphere.com© 2020 Labsphere, Inc. All Rights Reserved03© 2019 Labsphere,PB-13027-000
Inc. All Rights RevReserved
PB-13027-000 Rev 02

Ordering Information
Model Number:
Order Number:

FPT-600-050
AA-90041-050

FPT-600-100
AA-90041-100

FPT-610-050
AA-90043-050

FPT-610-100
AA-90043-100

Sphere:LMS-050

LMS-100

LMS-050

LMS-100

Spectrometer:

CDS-600

CDS-600

CDS-610

CDS-610Spectrally-Calibrated Lamp: (2π and 4π are not interchangeable)
2π
2PI-1-INT-650

2PI-1-INT-650

2PI-1-INT-650

2PI-1-INT-650Lamp Socket Assembly:

2π2π2πAbove Systems Include:

2π

Photometry Module: (includes Cube Computer)
Aux Lamp:

AUX-650

AUX-650

AUX-650

AUX-650

Software:FPT Software

FPT Software

FPT Software

FPT Software

min
0.05
0.04
0.03
0.05
0.05

min
0.20
0.16
0.13
0.20
0.12

All systems include Retouch 6080 and Tool Kit
FPT Replacement Parts
2PI-1-INT-650 2π Reference Lamp, internal mount - NVLAP*

AS-80003-100

AUX-650 Absorption Correction Lamp

AS-02986-650		

CDS-600 Mini Spectrometer

AS-02767-000

Extended Warranties		
Extended Warranty:

Software (12 Months)

AA-01375-000

Extended Warranty:

Parts Only (12 Months)
AA-01375-001(not including lamps)		

Extended Warranty:

Parts and Labor (12 Months)(not including lamps)

AA-01375-002*ISO/IEC 17025:2005 Lab Code 200951-0		

Performance Specifications (lumens)
Tungsten Filament:
Cool White LED:
Warm White LED:
Blue LED:
Red LED:
Upper Range:

min
0.05
0.04
0.03
0.05
0.05

max
7500
7100
4500
300
300

Ambient temp
can not exceed
100°C		

Advancing the Technology of Light: Measure. Create. Reflect.

min0.20
0.16
0.13
0.20
0.12max
16300
14500
13800
1200
1100

Ambient temp
can not exceed
100°C		

max
7500
7100
4500
300
300

Ambient temp
can not exceed
100°C		

sales@labsphere.com

www.labsphere.com

max
16300
14500
13800
1200
1100

Ambient temp
can not exceed
100°C© 2020 Labsphere, Inc. All Rights Reserved
PB-13027-000 Rev 03

Optional Accessories Ordering Information
4π Kit FPT-600-050, FPT-610-050		AA-90042-050
Enhances FPT-600-050-US for 4π measurements
Includes lamp holder, lamp post, connector box cables and tools. SCL-650 not included
4π Kit FPT-600-100, FPT-610-100		AA-90042-100
Enhances FPT-600-100-US for 4π measurements
Includes lamp holder, lamp post, connector box cables and tools. SCL-650 not included

SCL-650		
AS-01335-100
35W 4π Spectral Flux Standard - NVLAP*			

CSFS-650		
AS-01336-100
35W 4π Spectral Flux Standards, set of 3 - NVLAP*		

2PI-3-INT-650		
AS-80003-101
2π Reference Lamp Kit, internal mount, set of 3 - NVLAP*		FPT-FLH-050		
Universal Flashlight Holder for 2π port for 0.5m FPT		FPT-FLH-100		
Universal Flashlight Holder for 2π port for 1.0m FPT				

Advancing the Technology of Light: Measure. Create. Reflect.

sales@labsphere.com

www.labsphere.com© 2020 Labsphere, Inc. All Rights Reserved
PB-13027-000 Rev 03

System Spectrometer Specifications
Model Number:

CDS 600

CDS-610Detector:

2048 element Linear CCD

2048 element Linear CCD

Spectral Range: (spectrograph)

200 - 850 nm

350 - 1100 nm

Resolution: (FWHM)

1.5 nm

1.5 nmIntegration Time:

1 ms - 5 sec

1 ms - 5 sec

Cooling:

n/an/aLinearity:± 0.13%± 0.13%

Wavelength Accuracy:< ± 0.3 nm< ± 0.3 nm

Average % Noise on 100% Line:

0.20%

0.20%

Stray Light: (Y-50 filter)

1.66%

1.66%

Stray Light LED/Laser:

2.6E-5 from 450 - 550 nm w/633 nm laser

1.8E-5 from 450 - 550 nm w/633 nm laser

Optical Input:

600 um, 3 m long SMA Connection

600 um, 3 m long SMA Connection

Measurement Dynamic Range:

15K

15Kx, y Chromaticity Accuracy:<0.001 for x, y

<0.001 for x, y

Mechanical Shutter:

NoNo

AD Converter:

16 bit16 bitPC Interface:USB 2.0

USB 2.0Trigger: hardware

NoNo

Trigger: software

Yes

Yes

OD Filters:

NoNo

Shutter:

NoNo				 				NOTES:
1. Values above are the noise equivalent power in W/nm or lumens for the different wavelength ranges sited.
They were all taken with a 5W lamp, 10” sphere and 10 ms integration time.

Advancing the Technology of Light: Measure. Create. Reflect.

sales@labsphere.com

www.labsphere.com© 2020 Labsphere, Inc. All Rights Reserved
PB-13027-000 Rev 03

Integrating Sphere Specifications
Model Number:

LMS-050

LMS-100

Order Number:

AA-30000-050

AA-30000-100

Sphere Size:

50 cm

100 cmRequired Footprint:
closedopen

74.1 cm x 50.4 cm x 74.0 cm
103.7 cm x 89.9 cm x 74.0 cm

128.0 cm x 98.1 cm x 166.7 cm
174.4 cm x 165.9 cm x 166.7 cmSphere Weight:

20 Kg

92 Kg

Sphere Color:

Labsphere Blue

Labsphere Blue

Sphere Open Style:Clam Shell

Clam Shell

Sphere Assembly:

Spun Aluminum

Spun Aluminum

Frame Style:

Extruded Aluminum

Extruded Aluminum

Sphere Coating:Spectraflect

Spectraflect®

Spectraflect® Coating Reflectance:> 97% (nominal)> 97% (nominal)

Total Number of Ports:

66

SMA Adapter:

Included

Included

Cosine Corrector:

Included

IncludedExternal DUT 2π Port Size:

15 cm

33 cm

Detector Port Dimension:

1.25 cm

1.25 cm

Detector Port Quantity:

22Port Frame Reducer:

15 - 2.5 cm; 15 cm - 2.5 cm

33 cm - 15 cm; 15 cm - 2.5 cm

Temperature Probe Port:

2.5 cm

2.5 cm®

Max Recommended Lamp Size:<7 cm diameter, 33 cm long
<14 cm diameter, 67 cm long(LM-79) 4π geometry			
Max Port or DUT Area: (2π)

15 cm diameter; 71 cm233 cm diameter; 337 cm2

Max Recommended Linear DUT
Dimension: (2/3 sphere diameter)

33 cm

66 cm

Max Recommended Internal Surface Size:
I(2% Rule) (cm2)

50 cm2200 cm2

Maximum Sphere Coating Temp:

100°C

100°C

Advancing the Technology of Light: Measure. Create. Reflect.

sales@labsphere.com

www.labsphere.com© 2020 Labsphere, Inc. All Rights Reserved
PB-13027-000 Rev 03