HLV3 series

LED Spot Lights
HLV3 Series
HLV3-22-4S

HLV3-22-4M

HL2-25-P

HL2-50-P

HL2-M50-C

SE-65-M

HLV3-22-1 HLV3-22-2

SE-110-M

HLV3-22-2-1220

HLV3-22IR860
HLV3-22-4-NR

HLV3-14

HLV3-14-HU

PJ2-1505-2CA-PE

PJ2-3005-4CA-PE

HLV3-3M-RGB-4

Achieved both high output and high uniformity

Optimized for high-resolution telecentric lenses

Reduced individual differences in brightness

CCS Inc.

3rd Gen Lights with Exceedingly High Performance and Quality
Achieved both high output and high uniformity

Optimized for CCS’s high-resolution telecentric lenses

A rich lineup for a variety of applications.

Optimum Spot Lights for lens magnifications.

Reduced individual differences in brightness
Reduced maintenance labor times by controlling variations in
the brightness of Spot Lights.

Full Lineup with a Total of 29 Models
High Output Models HLV3-22-4S
High Uniformity Models HLV3-22-4M

HLV3-22-4S, a high output models
Optimized for CCS’s high-magniﬁ cation telecentric lenses
Imaging with a 6.0x lens

Spot Light (white)

HLV3-22SW-4S+

WD 64 mm

Telecentric Lens
6x, WD: 65 mm

SE-65VT60-M

Weight53g

HLV3-22-4M, a high uniformity models
Optimized for CCS’s low-magniﬁ cation telecentric lenses
Spot Light (white)

Imaging with a 0.3x lens

HLV3-22SW-4M
HLV3-22RD-4S
HLV3-22RD-4M

HLV3-22SW-4S
HLV3-22SW-4M

HLV3-22BL-4S
HLV3-22BL-4M

HLV3-22GR-4S
HLV3-22GR-4M

SE-110VT03-5M

WD 110.3 mm+

Telecentric Lens
0.3x, WD: 110 mm

Each color represents the brightness in steps of 10% the maximum value.
The results for individual products may vary.

Standard Models HLV3-22-1 HLV3-22-2
Housing has been redesigned and optimized from that ofconventional products.

Weight46

g27.1 mmHorizontal space saved by using an L-shaped housing.

HLV3-22RD-1
HLV3-22RD-2

HLV3-22SW-1
HLV3-22SW-2

HLV3-22BL-1
HLV3-22BL-2

HLV3-22GR-1
HLV3-22GR-2

19 mm shorter thanconventional products

Mounting holes on rear of Spot Light enable flexible installation.

IR Model HLV3-22IR860

HLV3-22-2-1220

Ø12-mm Light Emitting Tip

Infrared (860 nm)
models available

The HLV3-22-2-1220 series isderived from the HLV3-22-2
series. The main difference in
specifications is the diameterof the light emitting tip.

For transmission
applications.

WeightWeight

47

53

g

g

HLV3-22RD-2-1220 HLV3-22SW-2-1220 HLV3-22BL-2-1220 HLV3-22GR-2-1220

HLV3-22IR860

Compact and Lightweight Models HLV3-14 HLV3-14-HU
Both standard and high-uniformity types are available
Weight18g

HLV3-14RD
HLV3-14RD-HU

1

HLV3-14SW
HLV3-14SW-HU

HLV3-14BL
HLV3-14BL-HU

HLV3-14GR
HLV3-14GR-HU

Emitting surface of HLV3-14RD
Redlight
light
Red

Emitting surface of HLV3-14RD-HU
Redlight
light
Red

Comparison using our measurement conditions.
The data included is for reference only and the results for individual products may vary.

HLV3 Series

LED Spot Lights

Revamped Line of Lights to Satisfy Your Specific Needs
HLV3 Series Comparison Table
Model

Description

Input current(max.)

HLV3-22-4S

High output model

1,000 mA

HLV3-22-4M

High uniformity model

1,000 mA

HLV3-22-2

Standard model

700 mA

HLV3-22-1

Standard model

385 mA

HLV3-14Compact model with condensed light

275 mA

HLV3-14-HU

Compact model with diffused light

275 mA

HLV2-22-3W

Brightest model in HLV2 series(Reference data)

700 mA

Brightness (index)
RD (red)SW (white)

BL (blue)

GR (green)

1.6
1.0
1.1
0.6
1.0

1.9
1.3
1.5
0.9
0.8
0.1
1.0

2.0
1.2
1.5
0.9
1.3

1.6
1.0
1.2
0.8
1.1
0.1
1.0

x
x
x
x
x

< 0.1

x

1.0

x
x
x
x
x
x

x
x
x
x
x

< 0.1

x

1.0

x
x
x
x
x
x

Comparison using our measurement conditions. The data included is for reference only and the results may vary.

Reduced Individual Differences (Excluding HLV3-14, HLV3-14-HU, and HLV3-22IR860)
The newly designed HLV3-series Spot Light allows for brightness adjustment to the standard value.
In the production process, the brightness of each HLV3-series Spot Light is measured and adjusted to the standard value establishedfor each model. This allows us to produce high-quality Spot Lights with minimal variations in brightness between units of the same model.
The HLV3 series helps reduce labor time to install, maintain, and adjust brightness of Spot Lights and helps reduce time required on site.

Optimized for the Optical System of CCS High-Resolution Telecentric Lenses
Optimum Spot Lights for lens magnifications.
High uniformity types are available for low magnification lenses and high output types are available for high magnification lenses.
Spot Lights use an original lens to maintain high quality with minimal deviation in the optical axis.
Use Spot Lights with our original telecentric lenses (coaxial type) to create a stable imaging environment.

List of the recommended combination of telecentric lens and HLV3 series
Magnification
Telecentric lens

WD

0.3

65 mm
SE-110VT03-5M

110 mmHLV3 series

0.5

0.8

1.0

SE-65
VT05-M

SE-65
VT08-M

SE-65
VT10-M

SE-110VT05-M

SE-110VT08-M

SE-110VT10-M

HLV3-22-4M

1.5

2.0
SE-65
VT20-M

SE-110VT15-M

4.0

6.0

SE-65
VT40-M

SE-65
VT60-M

3.0

SE-110VT20-M

SE-110VT30-M

SE-110VT40-M

HLV3-22-1, HLV3-22-2, HLV3-22-4S, HLV3-14, and HLV3-14-HU
Recommendation based on the test results using our measurement conditions.

CCS High-Resolution Telecentric Lenses SE-65-M/110-M Series
WD 65 mm Type SE-65-M Series
0.5x

0.8x

1.0x

2.0x

4.0x

WD 110 mm Type SE-110-M Series
6.0x

0.3x

0.5x

0.8x

1.0x

1.5x

2.0x

3.0x

4.0x

Refer to the back cover for the detailed specifications.

2Lineup
LED color

Peak wavelength /correlated color
temperature (typ.)

HLV3-14RDRed

635 nm

0.7 W

HLV3-14SW

White

5000 K

0.9 W

HLV3-14BLBlue

465 nm

0.9 W

HLV3-14GR

Green

525 nm

HLV3-14RD-HU

Red

635 nm

HLV3-14SW-HU

White

4900 K

0.9 W

HLV3-14BL-HU

Blue

465 nm

0.9 W

HLV3-14GR-HU

Green

525 nm

1.1 W

HLV3-22RD-1

Red

630 nm

1.1 W

HLV3-22SW-1

White5600 K

HLV3-22BL-1

Blue

465 nm

HLV3-22GR-1

Green

520 nm

1.5 W

HLV3-22RD-2

Red

630 nm

2.1 W

HLV3-22SW-2

White5600 K

HLV3-22BL-2

Blue

465 nm

HLV3-22GR-2

Green

520 nm

2.8 W

HLV3-22RD-2-1220

Red

630 nm

2.1 W

HLV3-22SW-2-1220

White

5600 K

HLV3-22BL-2-1220

Blue

465 nm#Model name

1

2

3

4

Input current(max.)

275 mA385 mA

700 mA

700 mA

Power consumption(max.)

Extension cables

Options

1.1 W

Recommended
Control Units

Weight(max.)

18 g

−

0.7 W

PD3 *1FCB *2

1.4 W

Straight Cable

1.5 W

CC-PJ-0707

FRCB

PJ

Robot Cable

46 g

PJ2

Condenser Lens

2.5 W
2.7 W

2.5 W

HLV3-22GR-2-1220

Green

520 nm

HLV3-22RD-4M

Red

630 nm3.2 W

HLV3-22SW-4M

White

5600 K

3.7 W

HLV3-22BL-4M

Blue

465 nm

4.0 W

HLV3-22GR-4M

Green

520 nm

HLV3-22RD-4S

Red

630 nm

HLV3-22SW-4S

White

5600 K

3.7 W

HLV3-22BL-4S

Blue

465 nm

4.0 W

HLV3-22GR-4S

Green

520 nm

4.1 W

HLV3-22IR860

Infrared

860 nm

47 g

−

2.7 W
2.8 W

FCB *2

Straight Cable

4.1 W
1000 mA

FRCB

3.2 W

Robot Cable

Condenser Lens

PJ2

FCB-HLV3-10

53 g

Straight Cable

FRCB-HLV3-10
Robot Cable

4.1 W*1 The PD3-3024-3 and PD3-5024-3 series are not applicable to these products.

*2 The cables with a model name that ends with "-PF", "-PF-EL9", "-ME7", or "-EL2" are not included.

Common Specifications

Light Spectrum
100

1: Signal, 2: (+), 3: (-)

Cable length

300 mmCooling method

Natural air-cooling

Operating env. (indoors only)

Temperature: 0 to 40°C, Humidity: 20 to 85%RH (with no condensation)

Storage environment

Temperature: -20 to 60°C, Humidity: 20 to 85%RH (with no condensation)

CE marking

Safety standard: Conforms to EN62471

Cable material

Aluminum alloy

HLV3-14/-HU
465
nm

80
60

525 nm635 nm40

4900 K

20
0
400

100
Relative radiantintensity (%)

Polarity

Relative radiantintensity (%)

SMR-03V-B

Connector

450

500

550

600

5000 K
650
700
750
Wavelength (nm)

800

850

900

950

850

900

950

HLV3-22-1/-2/-2-1220/-4M/-4S, HLV3-22-IR860
465
nm

80
60

520 nm

630 nm

860 nm

40
20
0
400

5600 K
450

500

550

600

650
700
750
Wavelength (nm)800

Optional Cables (Sold Separately)
Extension
cablesStraight cables

Robot cables

FCB-1/-2/-3/-5(1 m / 2 m / 3 m / 5 m)

FRCB-1/-2/-3/-5(1 m / 2 m / 3 m / 5 m)

FCB-HLV3-10 (10 m) *FRCB-HLV3-10 (10 m) *The total length of the FCB-1/-2/-3/-5 and FRCB-1/-2/-3/-5

Notes

Extension Cables must be no longer than 5 m.
Branch cables are not available for the HLV3 Spot Lights.

* These 10-m extension cables connect the HLV3-series Spot Light and the PJ2-series Control Unit.

(60)

Weight: 1,200 g max., Cable permitted bending radius: 65 mm

3(Unit: mm)

10,000
(80)

(80)

(60)

(60)

(Ø4.3)(60)

Robot Cable(Ø7.8)(80)(Ø4.3)(Ø4.3)(Unit: mm)

FRCB-HLV3-10

10,000(80)(Ø8.1)

Straight Cable(Ø4.3)FCB-HLV3-10

Weight: 1,200 g max., Cable permitted bending radius: 35 mm

LED Spot Lights

HLV3 Series

Dimensions(Unit: mm)

HLV3-14RD/SW/BL/GR(4.4)

300 +35
0

1 HLV3-14RD-HU/SW-HU/BL-HU/GR-HU

Ø14 -0.1

7

0

-0.1

8 12
28

Ø8 -0.2

720

Ø7 (Emitting surface)

14

HLV3-22RD-1/SW-1/BL-1/GR-1

300 +35
0

2 HLV3-22RD-2/SW-2/BL-2/GR-2

(8)

(6)
Ø6 (Emitting surface)
For fixing options

2xM3x442.6

11

Ø14 -0.1

-0.02

8 12

11

0

7.6

11

Ø8 -0.10

11

14

Ø21.9

49

(For installation)

22

(8)

(6)

300 0

+35

3 HLV3-22RD-2-1220/SW-2-1220/BL-2-1220/GR-2-1220

Ø6 (Emitting surface)

20

7.6

11

42.6

-0.02
Ø12 -0.10

11

11

14

Ø21.9

49

2xM3x4(For installation)

11
22

HLV3-22RD-4M/SW-4M/BL-4M/GR-4M

4 HLV3-22RD-4S/SW-4S/BL-4S/GR-4S

(8)

(6)

300 +350

HLV3-22IR860

Ø6 (Emitting surface)
For fixing options

2xM3x5

0

8 12
46.6

Ø14 -0.1

7.6

-0.02

11

Ø8 -0.10

11

11

14

Ø21.9

49

(For installation)

11
22

4

Condenser Lenses

Installable on

HLV3-22-1, HLV3-22-2
HLV3-22-4M, HLV3-22-4S, HLV3-22-IR860

Select lenses according to illuminating distance, illuminating range, and uniform region to achieve optimum illumination, even in
imaging environments in which the workpiece is far away.

High Parallelism

HL2-50-P
HL2-25-P
Condenser lens for spot illumination
with high parallelism.

Compact condenser lens for spot
illumination with high parallelism.

Illuminance Distribution
HL2-25-P

HL2-50-P
50 to 100% illuminance range
5 to 100% illuminance range
0

50

100

200

300

500

1,000

1,500 (mm)Measured with HLV3-22SW-4S (white) mounted on the lens. Comparison using our measurement conditions. The data included is for reference only and the results for individual products may vary.

Illuminance Distributions (LWD Characteristics)
HL2-25-P + HLV3-22SW-4S(Unit of length used for the diameter of illuminated range: mm)

LWD

50 mm

100 mm

200 mm

300 mm

500 mm

1,000 mm1,500 mm

50 to 100% illuminance

Ø18

Ø15

Ø19

Ø28

Ø45

Ø89

Ø128

5 to 100% illuminance

Ø30

Ø30

Ø39

Ø48

Ø69

Ø141

Ø194

LWD

50 mm

100 mm

200 mm

300 mm

500 mm

1,000 mm

50 to 100% illuminance

Ø41

Ø40

Ø38

Ø35

Ø28

Ø44

Ø67

5 to 100% illuminance

Ø53

Ø52

Ø52

Ø54

Ø56

Ø79

Ø102

HL2-50-P + HLV3-22SW-4S(Unit of length used for the diameter of illuminated range: mm)

1,500 mm

Measured values, where the maximum illuminance within the range is 100%, using our measurement conditions. Results for individual products may vary.

Adjustable Focus
The length of the lens barrel varies, depending on the distance, to concentrate the light.

HL2-M50-C

50 to 100% illuminance range

Illuminance Distribution

5 to 100% illuminance range

HL2-M50-C

0

50

100

200

300

500

Measured with HLV3-22SW-4S (white) mounted on the lens. Comparison using our measurement conditions.
The data included is for reference only and the results for individual products may vary.

Illuminance Distributions (LWD Characteristics)
HL2-M50-C + HLV3-22SW-4S(Unit of length used for the diameter of illuminated range: mm)

LWD

50 mm

100 mm

200 mm

300 mm

500 mm

1,000 mm1,500 mm

50 to 100% illuminance

Ø5

Ø10

Ø19

Ø27

Ø46

-

-

5 to 100% illuminance

Ø18

Ø24

Ø38

Ø49

Ø75

-

-

Measured values, where the maximum illuminance within the range is 100%, using our measurement conditions. Results for individual products may vary.

5

HLV3 Series

LED Spot Lights

Irradiance Comparison
Irradiance graph (LWD characteristics)

Irradiance (μW/cm²)

350,000

HL2-25-P

300,000
250,000
200,000

HL2-50-P

150,000
100,000

HL2-M50-C

50,000
0

100

300

500

700

900

1,100

1,300

1,500

Illuminating distance (mm)Measurement with HLV3-22SW-4S (white) mounted on the lens. Measured values using our measurement conditions. Results for individual products may vary.

Condenser Lens Dimensions
2xM3(For mounting Spot Lights)

53

Ø32

Ø18

°

°
3
14

20

Surface A
75.3

(2.5)

Ø48
Condensing lens

90

90

31

Same as Surface A

Ø24
Condensing lensØ57

Same as Surface A

Ø14 +0.2
02xM3(For mounting Spot Lights)

HL2-50-P

Ø24

HL2-25-P(Unit: mm)

Ø14 +0.2
0M30.5xP0.5(For fixing options)

3
25

40

Surface A
(5.4)
125.9

M55xP0.75(For fixing options)

Weight: 85 g
5

10

2xM3x6(For installation)

Weight: 400 g

Accessories
Set screws (M3x3, black): 3 pcs
Hex wrench: 1 pc

10

20

2xM3x6(For installation)

Accessories
Set screws (M3x6, black): 3 pcs
Hex wrench: 1 pc

Polarizing filter (sold separately)
HL2-M50-C

Many sizes available for condenser lens.
Use in combination with another one installed on the camera lens.

M3(For mounting Spot Lights)

Ø48
Condensing lens

50

Ø57

Same as Surface A

6

Position adjusting screw(3.5)(83.5 to 108.8)

For HL2-50-P/HL2-M50-C Model: PL2-55-NL(with a screw to lock the polarizing direction)
M55xP0.75(For fixing options)

Attach the light aperture adapter to the HLV3 Spot Light to change thedirectivity of emitted light.
When combined with a condenser lens, use the light aperture adapter tomake fine adjustments to the illuminating distance and range.

18

3

Light aperture adapter (sold separately)

12
2xM3x6(For installation)

Accessories
Set screws (M3x6, black): 2 pcs
Hex wrench: 1 pc

Ø8 +0.1
0

Weight: 350 g

5Surface A

Model: PL-30/PL-30-NL(with a screw to lock the polarizing direction)Ø13.

Ø14 +0.2
0For HL2-25-P

7
13

2xM2
(For mounting
Spot Lights)

For SB-HL2-05: Ø0.5 mm
For SB-HL2-1: Ø1 mm
For SB-HL2-2: Ø2 mm(Aperture diameter)

Installable on■ HLV3-22-1/-2■ HLV3-22-4M/-4S
HLV3-22-IR860

Accessories
Set screws (M2x2, black): 3 pcs
Hex wrench: 1 pc

Put the Light Aperture Adapter and Condenser Lens all the way onto the tip of the Spot Light.
Light Aperture Adapter

Condenser Lens

HLV3-series Spot Light

6

Achieves optimum light color

Micro Fiber Head Dedicated Light Sources

HLV3-3M-RGB-4

HLV3-22-4-NR
HLV3-22SW-4-NR
HLV3-22RD-4-NR

1.5

x

2.3

Usage example
HFR-25-10 / HFR-25-30
HFR-40-20 (Three light sources are required for use.)x

That of Conventional
Products

Light Source
HLV3-3M-RGB-4

That of Conventional
Products

Dedicated Control Unit
(PJ2-3005-4CA-PE)
HFS-14-500

2.2

2.0

x

That of Conventional
Productsx

HLV3-22GR-4-NR

That of Conventional
Products

HLV3-3M-RGB-4

Usage example
HFR-25-10 / HFR-25-30
HFR-40-20 (Three light sources are required for use.)
Connecting adaptor(included with theMicrofiber Head)
HFS-14-500

HLV3-22BL-4-NR

Dedicated Control Unit
(PJ2 Series)

Light Source
HLV3-22-4-NR

Comparison with the conventional product HLV2-22-NR-3W of the same color.
Comparison using our measurement conditions. The data included is for reference only and the results may vary.
For detailed information on Micro Fiber Heads, refer to our website or the General Catalog of LED Lighting for
Machine Vision Applications.

Step-less Blending of Any Color with High Output and Uniformity
Illuminate with optimum red, green, and blue emission colors
HLV3-3M-RGB-4 is a dedicated light source composed of light sources and a blending
unit. Step-less intensity control of the light sources can be performed for each color. The
blending unit achieves uniform illumination due to its special construction. Connect thislight source to our unique Micro Fiber Head series to create optimum illumination colors ina variety of illumination configurations.

HLV3-22GR-4-NR(Green light source)

HLV3-22RD-4-NR

Connection example for
HLV3-3M-RGB-4 anda Micro Fiber Head

Straight type: HFS-14-500(Red light source)

HLV3-22BL-4-NR(Blue light source)

Ring type: HFR-25-10 / HFR-25-30

HFR-40-20 (Three light sources are required for use.)

HLV-3M (Blending unit)

Connect a Micro Fiber Head to easily
achieve full color illumination in a varietyof illumination configurations.

Change Emission Color to Extract Characteristics for Your Purpose
Example of imaging an LCD color filter
Use the HLV3-22-NR-4 series to achieve high-accuracy blended color illumination.
You can create the desired illumination color with independent intensity control of each light source to help improve inspection accuracy.
Red illumination

7

Blue illumination

Green illumination

Blend of green and blue illumination

Connecting the HLV3-22-4-NR Series and a Micro Fiber Head
Increase output by using HLV3-22-4-NR

Connection example for HLV3-22-4-NR and a Micro Fiber Head

Red (RD), green (GR), blue (BL), and white (SW) emission colors
are available so that emitted light close to that of a single-color LED
can be used for the spectral characteristics of the workpiece. You
can accurately extract characteristics when imaging workpieces bycombining the HLV3-22-4-NR with various micro fiber heads andselecting the optimum emission color and illumination configuration.

Straight type

Ring type

HFS-14-500

HFR-25-10 / HFR-25-30 / HFR-40-20(Three light sources are required for use.)

Easily Connect and Replace the Light Source
Instantly handle the selection of optimum emission color

The HLV3-22-4-NR series Micro Fiber Head dedicated light source can be easily installed and removed. Select the optimum emission
color when imaging workpieces to achieve accurate extraction of characteristics.
Specifications
Model namePeak wavelength /correlated color
temperature (typ.)LED color

Power consumption(max.)

Input current(max.)

Extension cables

Options

HLV3-22RD-4-NR

Red

630 nm

3.0 W

HLV3-22SW-4-NR

White

5,600 K

3.5 W

Straight Cable

HLV3-22BL-4-NR

Blue

465 nm

3.8 W

FRCB

HLV3-22GR-4-NR

Green

520 nmRed

630 nm

HLV3-3M-RGB-4

Blue

465 nmGreen

520 nm

1,000 mA

Recommended
Control Units

Weight(max.)PJ2

37 g

PJ2-3005-4CA-PE232 g

FCB *1

4.0 W

Robot Cable−

FCB-HLV3-10
Straight Cable

FRCB-HLV3-10

11 W

Robot Cable*1 The cables with a model name that ends with “-PF”, “-PF-EL9”, “-ME7”, or “-EL2” are not included.

Optional Cables (Sold Separately)
Extension
cablesStraight cables

Robot cables

FCB-1/-2/-3/-5(1 m / 2 m / 3 m / 5 m)

FRCB-1/-2/-3/-5(1 m / 2 m / 3 m / 5 m)

FCB-HLV3-10 (10 m) *FRCB-HLV3-10 (10 m) *The total length of the FCB-1/-2/-3/-5 and FRCB-1/-2/-3/-5
Extension Cables must be no longer than 5 m.

Notes

Branch cables are not available for the HLV3 Spot Lights.

* These 10-m extension cables connect the HLV3-series Spot Light and the PJ2-series Control Unit.

FRCB-HLV3-10(60)(Unit: mm)(Ø4.3)

(60)

10000
(80)

(Ø4.3)(Ø4.3)(Unit: mm)

Robot Cable(80)(Ø8.1)(80)

Weight: 1,200 g max., Cable permitted bending radius: 65 mm

(80)

(60)

(60)

(Ø4.3)

10000

Straight Cable(Ø7.8)FCB-HLV3-10

Weight: 1,200 g max., Cable permitted bending radius: 35 mm

Dimensions (mm)
HLV3-22RD-4-NR/SW-4-NR/BL-4-NR/GR-4-NR

27.5

HLV3-3M-RGB-4

27.5 14
HLV3-22RD-4-NR
HLV3-22GR-4-NR
HLV3-22BL-4-NR

3 0 0 +35

Ø220

3 0 0 +30 5

10
14
26

31
30
12

(1.2)

7 20

Ø10
(Emitting surface)

26.5

13

83
4xM3x4(For installation)

2xM3x5(For installation)

18

6

77

3

4

48.1

Ø14

2xM3x3(For installation)

11

14

2xØ3.5 Ø6 6(For installation)

Ø8(Emitting surface)

37

Ø22

0
Ø14 -0.1

42.1(18.4)

8

Dedicated Control Units for the HLV3/HLV2-Spot Lights Series
Installable on

All models of the HLV3/HLV2 Series

100 to 240 VAC
AC input type

Ethernet
ParallePJ2-1505-2CA-PE
2 channels

PJ2-3005-4CA-PE
4 channels

Easily Check and Set Operation of Spot Lights with the Built-in LCD
Analog Control Unit to which all HLV3/HLV2-series Spot Lights can be connected.
Light intensity can be set in 1024 steps. The built-in LCD allows for excellent reproducibility anddetailed control. Equipped with Ethernet and parallel communications for external control.
Available in 2-channel and 4-channel Spot Light output types. Select the type for your application.

Compliant models
PJ2-1505-2CA-PE
PJ2-3005-4CA-PE

Speciﬁcations
PJ2-1505-2CA-PE

Model name

PJ2-3005-4CA-PE

Lighting method

Continuous lighting

Drive method

Costant-current system

Intensity control method

Variable-current control

Number of channels

2 channels

Applicable Light Units

HLV3- and HLV2- series Spot Lights (The type of Spot Light is automatically detected.)

4 channels
5.5 VDC

Output voltage (ratings)

1,000 mA max./channel (Depends on the maximum input current of the Spot Light.)

Output current

100 to 240 VAC (+10%, -15%), 50/60 Hz

Input power
Power consumption

38 VA max.

68 VA max.
15 A (at 100 VAC), 36 A (at 240 VAC) from a cold start

Inrush current (typ.)3.5 mA max. (264 VAC, 60 Hz, with no load)

Ground leakage current

1500 VAC for one minute, Cutoff current: 10 mA,
500 VDC, 20 MΩ min.

Insulation withstand voltage(input-output, input-FG)
Overvoltage category

Category II

Operating environment

Temperature: 0 to 40°C, Humidity: 20% to 85% (with no condensation),
Altitude: 2,000 m max., Protective ground class: Class I, Pollution degree: 2, Indoor use only
Temperature: −20 to 60°C, Humidity: 20% to 85% (with no condensation)

Storage environment
Cooling method

Natural air cooling

CE marking

Safety standard: Conforms to EN 61010-1
EMC standard: Conforms to EN61000-3-2, EN61000-3-3, EN61000-6-2, and EN61000-6-4

Material, coating,and surface processing

Forced air cooling

Material: Aluminum and resin, Surface processing: Blue alumite

Weight

700 g max.
Instruction guide, 2-m-long 3-prong AC power cord with ground terminal

Accessories

Dimensions (mm)PJ2-1505-2CA-PE

LCD
Power switch
LAN connector
AC inlet
Trigger input terminal block

4xM3(Insertion depth: 5 mm)

9

130

82.6

Operatingknob

Parallelcommunications
connector
DIN railbracket

LCD

119.6

Outputconnectors
x2

Operating
knob

119.6

Parallelcommunications
connector

PJ2-3005-4CA-PE
130

Power switch

Outputconnectors
x4

LAN connector
AC inlet
Trigger input terminal block

4xM3(Insertion depth: 5 mm)

82.6

DIN railbracketDedicated Control Units for the HLV3/HLV2-Spot Lights Series
HLV3-14, HLV3-14-HU
HLV3-22-1, HLV3-22-2, HLV3-22-2-1220
All models of the HLV2 Series

Installable on

100 to 240 VAC

24 VDCAC input type

DC input type

PJ-1505-3CA
3 channels

PJ-1505-2CA
2 channels

PJ-1505-3CD24
3 channels

PJ-1505-2CD24
2 channels

Four Types to Match Your Use Environment
Dedicated Analog Control Units for HLV3/HLV2 Series

Compliant models

Step-less intensity control is performed by varying the current. There are 2 and3-channel Spot Light output types. You can select AC or DC power supply types.

PJ-1505-2CA, PJ-1505-3CA,
PJ-1505-2CD24, PJ-1505-3CD24

Digital Control Units
Installable on

HLV3-14, HLV3-14-HU
HLV3-22-1, HLV3-22-2, HLV3-22-2-1220
All models of the HLV2 Series

100 to 240 VAC

24 VDCAC input type

DC input type

Ethernet

Ethernet

EIA-485

ParalleParallePD3-5024-4 Series
4 channels

PD3-10024-8 Series
8 channels

PD3-5024-4 Series
4 channels

Select from Three Types of External Control for Your Network Environment
Digital Control Units selectable by control function
Compatible with two lighting modes: continuous lighting and ON/OFF lighting. Three types ofexternal control: Ethernet, EIA-485, and parallel. Available in 4-channel and 8-channel Light
Unit output types. Select the power supply from AC type and DC type.

Compliant models

PD3-5024-4 Series (AC input type)PD3-10024-8 Series

Dedicated Compact Controller for the HLV3/HLV2-Spot Lights Series
Installable on

32.5 mm 16 mm

Light intensity control
at every 10th level

24 VDCDC input type

70 mm

Perfect for narrow spaces,
used together with a Spot Lightto save space.

Light intensity control
at single unit level

HLV3-14, HLV3-14-HU
HLV3-22-1, HLV3-22-2, HLV3-22-2-1220
All models of the HLV2 Series

Setting the light intensity
x10

0

0

1

x1

0

1

1

9
9

Intensity level

1

2

12

100

Strobe mode

Selecting the strobe time
Constant lighting
ON/OFF mode
Spot Lights

HLV3-14
Series

50 μs, 100 μs, 250 μs, 500 μs,
1 ms, 4 ms, 10 ms, 40 ms
Response time: 50 μs max.
Overdrive is not supported.

A Single Unit Compatible with Continuous, ON/OFF, and Strobe Lighting
A compact, dedicated Controller for the HLV3/HLV2 seriesCompact size of only 16 mm × 70 mm × 32.5 mm (w×h×d) is optimal for installation in narrow spaces and forsaving space. Available in 1-channel Spot Light output type. Intensity value can be adjusted in 100 steps. Powersupply is 24 VDC, optimal for on-site usage.

Compliant

10

CCS High-Resolution Telecentric Lenses

SE-65-M Series / SE-110-M Series

Total of28

SE-65VT60-M
6x, WD: 65 mm

Speciﬁcations (SE-65-M Series)

ModelsModel name(Coaxial type)SE-65VT05-M

SE-65VT08-M

SE-65VT10-M

SE-65VT20-M

SE-65VT40-M

SE-65VT60-M

Model name(Straight type)SE-65ST05-M

SE-65ST08-M

SE-65ST10-M

SE-65ST20-M

SE-65ST40-M

SE-65ST60-M

Optical magnification

0.5x±5%0.8x±5%1.0x±5%2.0x±5%4.0x±5%6.0x±5%WD

65.1±2 mm

65.4±2 mm

65.1±2 mm

65.2±2 mm

65.3±2 mm

64±2 mmDepth of field *1*2

2.87 mm1.18 mm

0.79 mm0.26 mm0.08 mm0.06 mm

Resolution *2*3

12.0 μm

8.0 μm

6.7 μm

4.4 μm

2.9 μm

2.9 μm

NA*2

0.028

0.042

0.05

0.076

0.118

0.118

*2

Actual F-number (Fe)

9.0

9.4

9.8

12.9

16.9

25.5

TV distortion *2-0.001%+0.006%-0.023%+0.021%-0.003%-0.026%Weight (Coaxial)

80 g

62 g

54 g

55 g

85 g

100 g

Weight (Straight)

77 g

57 g

49 g

49 g

83 g

98 g

Mount

C mount

Maximum applicable image size

2/3 inch L (Ø12.75 mm)*4

2/3 inch (Ø11.00 mm)

Physical distance (O/I) *2

171.1 mm171.5 mm

161 mm160.9 mm

184.5 mm

216.7 mm

*1 The depth of field is a value calculated using 40 μm as the permissible circle of confusion.
*2 These are calculated values.
*3 The resolution is a value calculated using a 550 nm wavelength.
*4 2/3 inch L (Length: 5.98 mm, Width: 11.26 mm)
The specifications above are values based on the optical design. Differences between individual devices may occur due to assembly accuracy, etc.

SE-110VT03-5M
0.3x, WD: 110 mm

Speciﬁcations (SE-110-M Series)
5 M
Model name(Coaxial type)SE-110VT03-5M

SE-110VT05-M

SE-110VT08-M

SE-110VT10-M

SE-110VT15-M

SE-110VT20-M

SE-110VT30-M

SE-110VT40-M

Model name(Straight type)SE-110ST03-5M

SE-110ST05-M

SE-110ST08-M

SE-110ST10-M

SE-110ST15-M

SE-110ST20-M

SE-110ST30-M

SE-110ST40-M

Optical magnification

0.3x±5%0.5x±5%0.8x±5%1.0x±5%1.5x±5%2.0x±5%3.0x±5%4.0x±5%110.3±3.3 mm

110.8±3.3 mm

110.1±3.3 mm

110.5±3.3 mm

110.1±3.3 mm

110.6±3.3 mm

111.4±3.3 mm

110.5±3.3 mm

Depth of field *1*2

5.75 mm3.03 mm1.31 mm0.82 mm0.42 mm0.27 mm0.14 mm0.11 mm

Resolution *2*3

14.5 μm12.7 μm

8.8 μm

6.9 μm

5.2 μm

4.5 μm

3.6 μm

3.7 μm

NA*2

0.023

0.026

0.038

0.049

0.064

0.075

0.094

0.091

WD

*2

Actual F-number (Fe)

6.5

9.4

10.3

10.2

11.7

13.0

16.0

21.9

TV distortion *2+0.026%-0.009%-0.026%-0.0007%+0.024%-0.018%+0.008%-0.004%Weight (Coaxial)

212 g

131 g

106 g

108 g

110 g

115 g

166 g

121 g

Weight (Straight)

211 g

128 g

103 g

105 g

108 g

113 g

160 g

120 g

Mount

C mount

Maximum applicable image size

2/3 inch L*4(Ø12.75 mm)

2/3 inch(Ø11.00 mm)

268.2 mm259.7 mm

Physical distance (O/I) *2

2/3 inch L*4(Ø12.75 mm)
254.2 mm255.8 mm

256.9 mm

2/3 inch(Ø11.00 mm)
255.8 mm

272.9 mm

256.7 mm

*1 The depth of field is a value calculated using 40 μm as the permissible circle of confusion.
*2 These are calculated values.
*3 The resolution is a value calculated using a 550 nm wavelength.
*4 2/3 inch L (Length: 5.98 mm, Width: 11.26 mm)
The specifications above are values based on the optical design. Differences between individual devices may occur due to assembly accuracy, etc.

“CCS”, “LIGHTING SOLUTION”, and “HLV” are registered trademarks or trademarks of CCS Inc.

CAUTIONTo ensure proper and safe use of the product, please read the Instruction Guide completely before using the product.
The design and specifications of this product are subject to change without notification for product improvement.
The workpiece imaging examples included in this brochure are intended to serve only as references to help you select a suitable Light Unit. Please verify the functionality andconditions required for your particular application before you make a final selection. The sample workpieces used in this brochure have been processed specifically for sample
imaging. They are not intended to represent product quality and performance.

CCS Inc.

For information on your nearest CCS office, refer to our website.
http://www.ccs-grp.com/office/Headquarters (Kyoto, Japan)
TEL: +81-75-415-8284, FAX: +81-75-415-8316
E-mail: sales@ccs-inc.co.jp
http://www.ccs-grp.com/

CCS America, Inc. (USA)TEL: +1-781-272-6900, FAX: +1-781-272-6902
Email: info@ccsamerica.com
http://www.ccsamerica.com/

CCS Europe N. V. (Belgium)
TEL: +32-(0)2-333-0080, FAX: +32-(0)2-333-0081
Email: info@ccseu.com

CCS Asia PTE. LTD. (Singapore)
TEL: +65-6363-1180, FAX: +65-6363-1236
Email: sales@ccs-asia.com.sg
http://www.ccs-asia.com.sg/

CCS China Inc. (Shenzhen)
TEL: +86-755-8279-0477, FAX: +86-755-8279-0478
Email: ccschina@ccs-inc.co.jp
http://www.ccs-inc.cn/

Taiwan Office
TEL: +886-2-2581-7676, FAX: +886-2-2581-7662
Email: taiwan-tr@ccs-inc.co.jp

KOREA Testing Room
Email: ccskorea@ccs-inc.co.jp

Copyright © 2018 CCS Inc. All Rights Reserved.
Content current as of March 2018. 02002-00-1803-HLV3