Your local gas generation partner

Smaller. Simpler. Safer.
Streamline your GC with Precision SL hydrogen generator

Agas generation brandDiscover more at www.peakscientific.com/precisionSL

Precision Hydrogen SL)

99.9995% purity hydrogen gas on-demand)

Simple one-button start-up and shutdown)

Innovative sealed-capsule desiccant system, swap out
easily in 60 seconds)

Robust safety features including fail-safe auto-shutdown)

Optional auto-water fill or pressurized deionized water fill)

Safety compliant with CE certification)

2 year comprehensive warranty as standard

The smallest andeasiest to use hydrogengenerator for GC-FID.
Precision Hydrogen SL is everything we know
abouthydrogengeneration:

streamlined

toperfection for GC-FID. We’ve designed it to besmaller (less than a quarter the size of its
predecessor), child’s play to operate and maintainand above all, a safer alternative to having
compressed cylinders in your lab for GC flame
detector gas.

It’s all there in black and white.
They say the best things come in small packages. Not content with designing the smallest analyticalgrade hydrogen generator in its class, we set out to break the mould with a minimalist and streamlined
aesthetic. Available in a choice of black or white, Precision Hydrogen SL brings not only convenience andproductivity gains to your lab, but also a touch of style to match your GC instrumentation. If you thoughtusing a gas generator for your GC detector would bring complexity, cost or risk to your workflow, it’s timeto think again. Precision Hydrogen SL removes the worry of running out of gas, unreliable equipment,
expensive running costs or storing hazardous pressurized cylinders.

Smaller
Precision Hydrogen SL is one of the smallest laboratory grade hydrogengas generators available on the market, packing 100cc or 200cc flow ofhydrogen into a form that is less than 20% the size of its predecessor.

Simpler
Generating Hydrogen at the push of a single button, operation couldn’t
be easier. What’s more, replacing the proprietary desiccant capsule
periodically takes less than 60 seconds, no tools required.

Safe
Uncompromised safety in your laboratory. Unlike pressurized
cylinders, gas is generated on-demand with minimal gas stored andadvanced fail-safe technology.

Available in black or white and with two models to choose from - 100cc and 200cc.

Simplicity to use and maintain.
Precision Hydrogen SL is a lesson in streamlined simplicity. Gas is produced at the push of one button,
going into standby or power down with another push. It’s that easy. What’s more, periodic maintenance isjust as simple. An innovative sealed-capsule desiccant system can be swapped out and replaced as needed,without technical skills or tools, in under 60 seconds. The De-ioniser cartridge is a once annual service , againchanged easily in just a minute or two. No engineer visit necessary and virtually zero instrument down-time.
It takes longer to switch out empty gas cylinders.

A safer solution.
Hydrogen in the lab is always a source of concern. High pressure cylinders can be a risk to lab personnel
handling them as well as a large volume of gas that can leak into the lab environment. With Precision
Hydrogen SL in your lab, you eliminate these risks entirely. The generator produces gas only on-demand tomeet instrument need and stores minimal gas in the instrument (<150mL), with an automatic fail-safe to shut
off the generator in event of any leak.

Confidence and peace of mind.
As with all Peak Scientific gas generators, Precision Hydrogen SL is designed and engineered to the highest
quality standards in the UK and fully compliant with all safety standards including CE. You can be sure ofrobust and reliable performance with the backup of our industry-leading global technical support and fullycomprehensive 2 year manufacturer warranty, giving you uncompromised peace of mind.

Precision SL

Technical Specifications

Precision Hydrogen SL 100

Precision Hydrogen SL 200

Max Flow Rate
Max Pressure
PurityGas Outlets
Water Purity Requirement
Water Consumption
Operating Temperature
Electrical Requirements
Power Consumption
Heat Output
Generator Dimensions (HxWxD)
Generator Weight

100cc/min

200cc/min

100psi /6.9bar
99.9995%
1 x 1/8” Swagelok compression fitting
ASTM Type II (<1.0 uS / > 1MΩ)Up to 0.24L/day
Up to 0.24L/day
10°C - 35°C / 50°F - 95°F
100 - 240v ±10%/50/60 Hz/6A Max.
1.2A @240V, 220W
983 BTU/hr250 x 160 x 355 mm / 9.9 x 6.3 x 14’’8Kg( 17.7lbs)

Ordering Information
Manual Fill Part Number B/W

3302946 / 3302943

3302952 / 3302949

Auto-Water Fill Part Number B/W

3302947 / 3302944

3302953 / 3302950

Presurrized Water Fill Part Number B/W

3302948 / 3302945

3302954 / 3302951

Desiccant Capsule3302581De-ioniser Cartridge

3302583

Annual Service

www.peakscientific.com/ordering

Contact us today to discover more!
North America				Europe
Tel: +1 866 647 1649			
Tel: +44 141 812 8100
China						India
Tel: +86 21 5079 1190			Tel: 1 800 270 0946
Web: www.peakscientific.com/precisionSL Email: discover@peakscientific.com

Copyright 2023 Peak Scientific - Precision SL Series 2023 Rev 8 PN 3303377 RSID 3176 EN - 16/05/2023