Genius 1051

Nitrogen & Air Generator for Shimadzu LC-MS
Part Number : 10-9558

Service Kit : 08-4824

Your local gas generation partner

Description
Engineered specifically for the Shimadzu LCMS-8045, LCMS-8050
and LCMS-8060 instruments, the Genius 1051 provides sufficient
nitrogen and dry air when these instruments are operated in“standard gas delivery mode”. If operating in “high flow gas
delivery mode” the Genius 3051 is the most suitable gas generator
solution.
With one output of phthalate free, high purity nitrogen and one
output of dry air, this tailored solution can deliver any combinationof gases up to a total of 25L/min and has been tested andapproved by Shimadzu for the LCMS-8045, LCMS-8050 andLCMS-8060.
The Genius 10 series models are compressor based systems Peakmanufacture utilising Pressure Swing Adsorption and Carbon
Molecular Sieve to ensure high performance and supply of
analytical instrument grade nitrogen.

Applications))

1 x Shimadzu LCMS-8045 in standard gas delivery mode))

1 x Shimadzu LCMS-8050 in standard gas delivery mode))

1 x Shimadzu LCMS-8060 in standard gas delivery mode

Key Features

600.00 mmOFFON))

Shimadzu approved gas solution for LCMS-8050 and LCMS8060))

Self-contained solution with integrated compressors, no needfor an external air supply))

Highly economical source of nitrogen/dry air with low lifetimerunning costs))

Latest generation high performance compressors for longerservice interval))

Easy to use, gas at the push of a button))

Wheels make it easy to manoeuvre generator around the lab))

12 month comprehensive on-site warranty

DRY AIR713.00 mm

750.00 mm

713.00 mm

750.00 mm

600.00 mm

Genius 1051 Dimensions

Copyright 2016 Peak Scientific - Rev - Genius 1051 - Rev 4 - 08/11/16

Technical Specifications

Genius 1051

Max Flow Rateup to 25 L/min /0.88 CFM of Air & Nitrogen combined

Max Pressure

6.9 bar / 100 psi

Max Relative Humidity

80% Non-Condensing

Max Altitude2000 Metres

Particles< 0.01μm

Gas Outlets

2x 1/4” BSPP

Phthalates

None

Suspended Liquids

None

Operating Temperature

5ºC - 35ºC / 41ºF - 95ºF

Electrical Requirements

230v 50/60Hz 7.0A

Power Consumption

1610 Watts

Generator Dimensions (HxWxD)
Generator Weight713 x 600 x 750 mm/28.1 x 23.6 x 29.5 in
118.5 Kg (261 lbs)

Ordering Information
Part Number

10-9558

Annual Service08-4824

Standard Maintenance Plan

09-3110Complete Maintenance Plan

09-3010

Step-Up Transformer For Use With
208V Electrical Supply

06-3200

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2016 Peak Scientific - Rev - Genius 1051 - Rev 4 - 08/11/16