Infinity 1031 - 1034
Nitrogen/Dry Air Generator for
PartNumber: 99-9999
Service Kit : 99-9999
SciexMassSpectrometers

Your local gas generation partner

Description
Membrane technology is at the heart of the Infinity 1031-1034 generators.
Delivering one output of laboratory-grade nitrogen,and two dry air outputs,
the Infinity 1031-1034 generators are a durable, hassle-free solution for
Sciex LC-MS instruments. The choice of Infinity model will depend on theapplication, with different gas flows defining the differing models.
Peak Scientific’s Infinity series generators have been engineered to provide
nitrogen to laboratories which already have an external source of clean,
dry air (we can provide assistance on external compressors or pre-filtration
systems as part of an Infinity gas generation solution). These systems
contain no moving parts, are effectively silent whilst in operation andrequire minimal maintenance.

Applications)

Nitrogen supply suitable formost LC-MS instruments

Key Features

Copyright 2020 Peak Scientific - Rev - Infinity 1031 - 1034 Data sheet Rev 7 RSID 405 EN - 02/03/20)

Capable of supplying multiple Sciex LC-MS
instruments)

24/7 operation at optimum performance if required)

Gas is supplied on demand so generator works to yourschedule)

Few moving parts means lower maintenancerequirement and ensures long life of the generator)

Minimal set-up required)

Completely silent in operation)

12 month comprehensive on-site warranty

Infinity 1031
(Hi Flow)

Infinity 1031

Infinity 1032

Infinity 1033

Infinity 1034

Max Flow Rate (Nitrogen)

10 L/min19 L/min38 L/min

57 L/min

76 L/minMax Flow Rate (Source Gas)

22 L/min

26 L/min

52 L/min78 L/min

104 L/min

Max Flow Rate (Exhaust Gas)

10 L/min

25 L/min

50 L/min

75 L/min

100 L/min

340 L/min

475 L/minTechnical Specifications

Max Gas Outlet Pressure

110 psi

Min Air Inlet Flow Rate

115 L/min

115 L/min

Min/Max Air Inlet Pressure

240 L/min
8.27 - 10.0 bar / 120 - 145 psi

Max Pressure Drop

8 psi

Min/ Max Operating Temperature

5ºC - 30ºC / 41ºF - 86ºF

Particles< 0.01 μm

Suspended Liquids

None

Electrical Requirements

None

Generator Dimensions (HxWxD)

760 x 371 x 200 mm/ 29.9 x 14.5 x 7.9 in

760 x 371 x 200 mm/ 29.9 x 14.5 x 7.9 in

730 x 428 x 250 mm/ 28.7 x 9.8 x 16.8 in

730 x 428 x 250 mm/ 29 x 12 x 29 in

960 x 369 x 250 mm/ 37.8 x 14.5 x 10 inGenerator Weight

14 kg (30.8 lbs)

14 kg (30.8 lbs)

21Kg (46 lbs)

27Kg (60 lbs)

27Kg (60 lbs)

﻿ 13-1280

﻿ 13-1270

13-1272

13-1274

13-1276

Ordering Information
Part Number
Annual Service﻿ 08-0317

Standard Maintenance Plan

Visit: www.peakscientific.com/ordering

Complete Maintenance Plan

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com
Copyright © Peak Scientific Instruments Ltd 2020 — Infinity 1031 - 1034 Data sheet Rev 7 RSID 405 EN —02/03/2020