The CondenSyn®& CondenSyn MINI
Waterless Reflux Condensers

Benefits:

Simple, safe and no risk of floodingRobustEasy to cleanHigh performance

Simple and safe to useEnvironmentally friendly

Short pay-back timeNo risk of flooding

Available in the 3 lengths to suit
all R&D scales, all with standard
socket sizes

Save money, save water, save the planet!
The smart solution for everyone
The average rate of water used in a condenser is 2 litres per minute.
The water costs used below are at £1.22 per cubic metre supply and£1.56 per cubic metre waste which is equivalent to 0.28p per litre.
This commercial rate cost is from a water supplier here in the UK,Anglian Water in March 2020.

NEW adapter available
to allow the CondenSyn
to be used for distillation
See details overleaf

See table (below) for savings.
Water used per day
Cost per day
Cost per month (working days)
Cost per year (working days)

5 hours per day
600 litres£1.68£33.60£403.20

24 hours per day
2880 litres£8.06£161.20£1,934.40

Ask for a demonstration online/in your laboratory today!

enquiries@asynt.com+44 (0)1638 781709

www.Asynt.com

CondenSyn & CondenSyn MINI
Independent tests were performed by a leading UK University to evaluate
the performance for safe use in their research and teaching laboratories.
Evaluations were based upon a basic 350 mm effective length CondenSyn
and a 250 mL round bottom flask with 150 mL of solvent.
Solvent
bp [°C]oil bath [°C]
difference [°C]time [min]%-loss (total)*%-loss per hour

DCM
4050
10240
-0.8%
-0.2%

Acetone
5671
15360
-1.3%
-0.2%

THF667812300-1.4%-0.3%

Ethanol
78
100
22300
-0.5%
-0.1%

Acetonitrile
82100
18300
-0.9%
-0.2%

Water100
120
20240
-1.6%
-0.4%

Toluene110125
15300
-0.9%
-0.2%

*It is recommended that to prevent further losses a temperature differential is
kept below 10 oC, especially when using solvents below 60 oC boiling point .

“Since receiving the CondenSyn waterless condenser, we have substituted standard
water condensers for the waterless counterpart in a number of reactions, including
overnight refluxes and Soxhlet extractions. We have found the transition toCondenSyn seamless! In our tests, CondenSyn has been as efficient as traditionalcondensers, however, the waterless setup is safer, greener and easier to use
without loss of performance. The lack of tubing also allows for a tidier and safer
working area and the sleek shape of the condensers would allow for multiple
condensers to be setup on the same hot plate, vastly enhancing productivity!

“Dr Panagiotis Manesiotis Research Team, Queen’s University Belfast

Popular Purchase Options
Key sizes shown below but other sizes are also available
200 mm recommended for use with flasks up to 100 mL in size†
350 mm recommended for use with flasks up to 500 mL in size†
450 mm recommended for use with flasks up to 1000 mL in size†
GB-C-200-B14 200 mm with B14 socket
GB-C-200-B19 200 mm with B19 socket
GB-C-200-B24 200 mm with B24 socket
GB-C-350-B14 350 mm with B14 socket
GB-C-350-B19 350 mm with B19 socket
GB-C-350-B24 350 mm with B24 socket
GB-C-350-B29 350 mm with B29 socket

enquiries@asynt.com

GB-C-450-B14 450 mm with B14 socket
GB-C-450-B19 450 mm with B19 socket
GB-C-450-B24 450 mm with B24 socket
GB-C-450-B29 450 mm with B29 socket
GB-C-DIST-B24 Distillation adapter for use with B24 socket CondenSyn
GB-C-DIST-THERM Thermometer and adapter set with B24 socket† Not more than half full.

+44 (0)1638 781709

www.Asynt.com