MEASURE
any light source

illumia Pro2-UV
UV-LED Characterization®

Accurately Characterize Packaged UV LEDs
UV LED performance depends on the junctiontemperature. Thermal variances at the junction
can impact UV LED output and life expectancy.
With Labsphere’s illumia®Pro2-UV users canquickly and accurately test UV-LED performance
as a function of thermal condition.
Dependable Results• NMI-Traceable Calibrated Xe
spectral radiant flux standard• High dynamic range for a varietyof light levels• Spectralon® integrating sphere,
EPV Spectralon optional• CDS-2600-UV Spectrometer withhighly-efficient stray light rejection

Advancing the Technology of Light: Measure. Create. Reflect.

Measure• Total Radiant Flux
• Total Photon Flux
• Electrical Power• Wavelength Characterization• Peak Wavelength• FWHM• L, I, V, T Sweeps
Applications• Germicidal UV (GUV)• UVC disinfection and purification• UV Curing• Medical phototherapy• Analytical instruments• Horticulture lighting

sales@labsphere.com

www.labsphere.com© 2020 Labsphere, Inc. All Rights Reserved
PB-14096-000 Rev 01

LIVT Sweep Measurement Functions
NameConstant

VaryMeasureILVTI

L, V

VLI

TVL, ITLVITL, V

TLI

VT

L, I

ILV/T

T for each
I Setting

I, TL, V

T for each
V setting

V, T

L, I		
VLI/T

Integral® Software

Measurement Parameters
Electrical: Current, Voltage, Electrical Watts
Optical: Spectral and Total Radiant Flux, Photon Flux,
Peak Wavelength, Center Wavelength,
Centroid Wavelength, FWHM
Thermal: Case Temperature Control vs. Electrical andOptical Parameters

Integral software is a comprehensive light test application
package. It allows for data collection and system control
of a variety of system configurations and applications.
As a certified National Instruments LabVIEW Alliance partner,
Labsphere has designed Integral to include robust reporting
capabilities. Integral includes multi-language support andcan be accessed remotely via an HTML5-enabled browser.
Integral also offers an optional API license option allowing
users to create their own programs and interface withexisting software applications.

Typical illumia®Pro2-UVSpecifications

System Spectrometer
Specifications

Measurement Range:
LED Optical Flux:
5 W Thermal Load Operating T:
Sphere Size:
Sphere Material:
Spectrometer:
Sourcemeter:
TE Chiller:
TEC Source:
Software:Spectrometer:
Detector:		
Key: I=current, L=optical watts, V=voltage, T=temperature

200 - 400 nm
1 mW - 2000 mW
20 - 85 C
6”Spectralon
CDS-2600-UV
Keithley 2400
Arroyo TE Chiller 207
Arroyo 5305
Integral

Cooling:
Spectral Range:
UV Calibrated Range:
Resolution:
Wavelength Accuracy:
Data Point Interval:
Integration Time
Dynamic Range:
Average % Noise: (360 - 830 nm)
Software Corrected Stray Light:

CDS-2600-UVTE Cooled 1044 x 64
CCD (back thinned)-10 ± 0.05 C
200 - 960 nm
200 nm - 400 nm
2.2 nm< ± 0.4 nm
1.0 nm8 ms - 900 Seconds> 200,000:1*
0.07%< 1.0%*** Measured as the saturation signal divided by the standard deviationof the dark signal with 10 scans averaged.
** Stray light is the average reported transmittance from 210 - 370 nm
through a 500 nm cut-on filter
Labsphere has created thousands of customized solutions, tailored to the specific and often proprietary needs ofour customers. Our expert team of Scientists and Engineers continue to advance our nearly 40 years of experiencein dozens of industries – and our Operations team quickly turns these into finished product through our internal
prototype shop and advanced manufacturing techniques. Contact Labsphere with your specialized needs today.

Advancing the Technology of Light: Measure. Create. Reflect.

sales@labsphere.com

www.labsphere.com© 2020 Labsphere, Inc. All Rights Reserved
PB-14096-000 Rev 01