Genius NM32LA

Nitrogen Generator for LC-MS
Part Number : 10-6020

Service Kit : See Reverse

Your local gas generation partner

Description
The Genius NM32LA uses tried and tested membrane technology
and internal compressors to generate laboratory grade nitrogenon-site and on demand. Built to last, with minimal maintenancerequirements, this generator has been designed for ease of use,
it will fit under a standard lab bench and is fitted with wheels foreasy mobility.
The culmination of over a decade’s work perfecting an on-site
nitrogen gas generation for LC-MS, the Genius NM32LA is thepinnacle of our Genius series. The NM32LA has become a provenand reliable on-demand nitrogen solution in countless laboratories
worldwide. This system’s popularity is underpinned by its
compatibility with most mass specs available in the market.

Applications)

Suitable for the majority ofLC-MS instruments

Key Features

600.00 mm713.00 mm)

Compressor based solution, generator can run on its own
without the need for an external source of air)Proven and reliable with 1000’s in the field worldwide)

Duty and service indicators)

Latest generation, high specification compressor located ininsulated chamber to minimize noise and vibration)

Once connected to the instrument and power is supplied, thesystem is fully operational)

The most cost eﬀective method of gas supply)

Remove the hassle and safety concerns of ordering andchanging out pressurized cylinders)

Gas is supplied on demand so generator works to your schedule)

Dual tap transformer available for labs with 110V electricalsupply750.00 mm

713.00 mm

Genius Nm32la Dimensions

Copyright 2021 Peak Scientific - Rev - Genius NM32LA EN Rev 10 RSID 487 - 04/01/21

Technical Specifications

Genius NM32LA

Max Flow Rate

32 L/minMax Pressure

100 psi / 6.9 bar

Max Relative Humidity

80% Non-Condensing

Max Altitude2000 MetresParticles< 0.01μm

Gas Outlets

1x 1/4” BSP Female

Phthalates

None

Suspended Liquids

None

Operating Temperature

5ºC - 35ºC / 41ºF - 95ºF

Electrical Requirements

230v 50/60Hz 7.0A

Power Consumption

1610 Watts

Noise Level

59 dB

Generator Dimensions (HxWxD)
Generator Weight713 x 600 x 750 mm/28.1 x 23.6 x 29.5 in
102.5 Kg (226 lbs)

Ordering Information
Part Number

10-6020

Annual Service
Standard Maintenance Planvisit: www.peakscientific.com/ordering

Complete Maintenance Plan
Step-Up Transformer For Use With
208V Electrical Supply

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2015

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2021 Peak Scientific - Rev - Genius NM32LA EN Rev 10 RSID 487 - 04/01/21