Create any spectrum

Albedo Sunlight Levels and Halogen
Spectrums for Full Testing Solutions
HELIOSPlus-AMA A FAMILY SYSTEMS

Value:
Great solution for “Test As You Fly” applications
The HELIOSPlus-AMA, A Family systems are the ultimate
expression of uniform calibration source technology
allowing you to test with spectrums and levels that areequivalent to solar outputs. Labsphere has
employed Xenon sources for bluer spectrum formany years, but in the A Family we have added
cutting edge Electrode-less Plasma lamps thathave incredible lifetime, light level outputs andradiometric stability. The A Family allows you to
confirm all your testing with classic QTH andaugment your characterization to test as you fly
in with traceable solar-like light.

• Great solution for R&D and laboratory
test flexibility• Included CCD spectrometer and systems
are compatible with Labsphere’s full family of
CDS spectrometer technology• All options in HELIOSPlus available for A Family
inclusion: Easy to configure a system to meetyour exact requirements• High level of absolute characterization

Performance:• Test As You Fly: Solar and QTH spectrums
from 250 - 2500 nm- Plasma 3000K to 5300K adjustable
CCT and spectrum- Xenon 3000K to 6500K adjustable
CCT and spectrum• Large Dynamic Light Range:>Albedo 0 (AM 0) down to night vision light levels- Cameras and sensors >16bit,
actual 23bit (150dB)- Resolution and adjustability tomeet customer’s application andbudget needs• UV (<380 nm) ranges available

www.labsphere.com | sales@labsphere.com© 2022 Labsphere, Inc. All Rights Reserved PB-14017-000 Rev 11

A Family: Air Mass (Albedo) Sunlight Source						Model Number
Smart Part Number

USLR-A20F-XAN2-P
USLR-A20F-XDN2-P
USLR-A20F-XMN2-P
A5XA-N2NN-NNAR-NS00-0000-P A5XD-N2NN-NNAR-NS00-0000-P A5XM-N2NN-NNAR-NS00-0000-P

OPTICAL PERFORMANCE SPECIFICATIONSUNIFORMITY (EACH LAMP OR COMBINED)				
Spatial Luminance Uniformity over Exit Port (f/4) - All Lamps On+/-1.0%+/-1.0%+/-1.0%Angular Uniform FOV (Full Angle) - Degrees / F# / NA - All Lamps On+/-2.0% - 35° / 0.85 / 0.6+/-2.0% - 35° / 0.85 / 0.6+/-2.0% - 35° / 0.85 / 0.6
QTH ONLY				
Expected Luminance Output: cd/m2
3,800
3,800
3,800
Expected Illuminance at Port: lux
12,00012,00012,000Est. Peak Radiance: W/m2-sr-um @ 0.95 um
100
100
100
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.95um
4.70E+20
4.70E+20
4.70E+20
Minimum Resolution: lux
1.20E-03
1.09E+00
2.40E-01
Approximate Correlated Color Temperature (QTH)
3000K +/-50K
3000K +/-50K
3000K +/-50K
Typical Lamp Lifetimes (hrs)>500hrs>500hrs>500hrs
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-10% & +/-200K-10% & +/-200K-10% & +/-200K
Est. Output Degradation over 50hrs (% & CCT Shift)-1.0% & +/-20K-1.0% & +/-20K-1.0% & +/-20K
PLASMA OR XENON ONLY
Expected Luminance Output: cd/m2
Expected Illuminance at Port: lux
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl or 0.47 Xe
Minimum Resolution: lux
Approximate Correlated Color Temperature
Plasma Rest Mode/Xenon Flicker (Peak-Peak/RMS/Hz) - Typical
Plasma Test Mode (Peak-Peak/RMS/Hz) - 30 minutes (Rest 5min)
Installed Cold Mirror (330-750nm)
Typical Lamp Lifetimes (hrs) Plasma/Xenon
Est. Lamp Degradation Over Lifetime (% & CCT Shift)Est. Output Degradation over 50hrs (% & CCT Shift)

16,50051,000210
6.30E+20
5.10E-03
6000K +200K/-400K
4%/1%/5kHz
N/A
Yes
400
-50% & -600K-2.5% / -30K

16,50051,000210
6.30E+20
4.64E+00
6000K +200K/-400K
4%/1%/5kHz
N/A
Yes
400
-50% & -600K-2.5% / -30K

16,50051,000210
6.30E+20
1.02E-00
6000K +200K/-400K
4%/1%/5kHz
N/A
Yes
400
-50% & -600K-2.5% / -30K

PLASMA/XENON & QTH (Both Full Open VA)
Expected Luminance Output: cd/m2
Expected Illuminance at Port: lux
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl @ 0.47 Xe
Approximate Correlated Color Temperature (Xenon & QTH)

20,00063,000280
9.00E+20
5100K +400K/-200K

20,00063,000280
9.00E+20
5100K +400K/-200K

20,00063,000280
9.00E+20
5100K +400K/-200K

ATTENUATORS
Number of Steps in Attenuator Range
Dynamic Range/Bits/dB - Full Range of System (both lamps)

2.00E+06
4.41E+07/25/152

1.20E+04
4.85E+04/15/93

1.00E+04
2.20E+05/17/106

INTEGRATING SPHERE
Coating / Material
Sphere Internal Diameter: Inches (Meters)
Frame Type
Output Port Size: Inches (Meters)

Spectraflect®
20 (0.5)
20 in Cage
8 (0.2)

Spectraflect
20 (0.5)
20 in Cage
8 (0.2)

Spectraflect
20 (0.5)
20 in Cage
8 (0.2)SYSTEM COMPONENTS
QTH Lamps Internal (# , Wattage)
QTH Lamps External (#, Wattage)
Xenon Lamp & Housing
Plasma Lamp & Housing
Power Supplies (# - Model)
Special Power Supply
Variable Attenuator
Monitor Detector(s)Detector Filters (in Filter Holder)
System Software
Spectral Radiance Monitor (Type, Spectral Range)(0)(1) 150300W
None(1) LPS-350
Integrated Xenon(2) VAA-220A
SD-S1
Photopic
HELIOSense
CDS-800(0)(1) 150300W
None(1) LPS-350
Integrated Xenon(2) VAD-012
SD-S1
Photopic
HELIOSense
CDS-800(0)(1) 150300W
None(1) LPS-350
Integrated Xenon(2) VA-MM
SD-S1
Photopic
HELIOSense
CDS-800

STANDARD SYSTEM CALIBRATIONS (NIST Traceable)				
Luminance
Yes
Yes
Yes
Correlated Color Temp (All lamps matched & w/VA position)
Yes
Yes
Yes
QTH Only Spectral Radiance (350-2400nm)
Yes
Yes
Yes
Xenon Only Spectral Radiance (350-2400nm)
Yes
Yes
Yes
QTH & Xenon Spectral Radiance (350-2400nm) & CDS
Yes
Yes
Yes
Exit Port Spatial Uniformity
Yes
Yes
Yes
Exit Port Angular Uniformity
Yes
Yes
Yes
Operational Duration of Calibration (Xe/QTH)
25 hrs / 50 hrs
25 hrs / 50 hrs
25 hrs / 50 hrs

www.labsphere.com | sales@labsphere.com© 2022 Labsphere, Inc. All Rights Reserved PB-14071-000 Rev 11

A Family: Air Mass (Albedo) Sunlight Source						Model Number
Smart Part Number

USLR-A12F-XAN2-P
USLR-A12L-XAN2-P
USLR-A12L-UAN1-P
A3XA-N2NN-NNAR-NS00-0000-P A4XA-N2NN-NNAR-NS00-0000-P A4UA-N1NN-NNAR-NS00-0000-P

OPTICAL PERFORMANCE SPECIFICATIONS			UNIFORMITY (EACH LAMP OR COMBINED)			
Spatial Luminance Uniformity over Exit Port (f/4) - All Lamps On+/-1.0%+/-1.0%Angular Uniform FOV (Full Angle) - Degrees / F# / NA - All Lamps On+/-2.0% - 35° / 0.85 / 0.6+/-2.0% - 35° / 0.85 / 0.6+/-1.0%+/-2.0% - 35° / 0.85 / 0.6

QTH ONLY			
Expected Luminance Output: cd/m2
11,50015,50015,500
Expected Illuminance at Port: lux
36,00049,00049,000Est. Peak Radiance: W/m2-sr-um @ 0.95 um
350
460
460
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.95um
1.50E+21
2.20E+21
2.20E+21
Minimum Resolution: lux
3.60E-03
3.60E-03
3.60E-03
Approximate Correlated Color Temperature (QTH)
3000K +/-50K
3000K +/-50K
3000K +/-50K
Typical Lamp Lifetimes (hrs)>500hrs>500hrs>500hrs
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-10% & +/-200K-10% & +/-200K-10% & +/-200K
Est. Output Degradation over 50hrs (% & CCT Shift)-1.0% & +/-20K-1.0% & +/-20K-1.0% & +/-20K
PLASMA OR XENON ONLY			
Expected Luminance Output: cd/m2
30,50042,00013,500
Expected Illuminance at Port: lux
96,000132,000
42,000Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
420
550
170
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl or 0.47 Xe 1.20E+20
1.60E+21
5.10E+20
Minimum Resolution: lux
9.60E-03
1.32E-02
4.20E-03
Approximate Correlated Color Temperature
6000K +200K/-400K
6000K +200K/-400K
6000 +/-300K
Plasma Rest Mode/Xenon Flicker (Peak-Peak/RMS/Hz) - Typical
4%/1%/5kHz
4%/1%/5kHz
4%/1%/5kHz
Plasma Test Mode (Peak-Peak/RMS/Hz) - 30 minutes (Rest 5min)
N/A
N/A
N/A
Installed Cold Mirror (330-750nm)
Yes
Yes
None
Typical Lamp Lifetimes (hrs) Plasma/Xenon
400
400
400
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-50% & -600K-50% & -600K-50% & -600K
Est. Output Degradation over 50hrs (% & CCT Shift)-2.5% / -30K
-2.5% / -30K
-2.5% / -30K
PLASMA/XENON & QTH (Both Full Open VA)				
Expected Luminance Output: cd/m2
37,50047,30032,500
Expected Illuminance at Port: lux
117,800
148,200
102,000
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
600
770
620
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl @ 0.47 Xe 1.14E+21
1.52E+21
1.03E+21
Approximate Correlated Color Temperature (Xenon & QTH)
5100K +400K/-200K
5100K +400K/-200K
5400K +400K/-200K
ATTENUATORS			
Number of Steps in Attenuator Range
2.00E+06
2.00E+06
2.00E+06
Dynamic Range/Bits/dB - Full Range of System (both lamps)
2.68E+07/24/148
3.37E+07/25/150
2.32E+07/24/146
INTEGRATING SPHERE			
Coating / Material
Spectraflect
Spectralon®
Sphere Internal Diameter: Inches (Meters)
12 (0.3)
11.5 (0.29)
Frame Type
12 in Cage
12 in Cage
Output Port Size: Inches (Meters)
4 (0.1)4 (0.1)

Spectralon
11.5 (0.29)
12 in Cage
4 (0.1)SYSTEM COMPONENTS			
QTH Lamps Internal (# , Wattage)
(0)
(0)
(0)
QTH Lamps External (#, Wattage)(1) 150(1) 150(1) 150
Xenon Lamp & Housing
175W
175W
UV 175W
Plasma Lamp & Housing
None
None
None
Power Supplies (# - Model)(1) LPS-350(1) LPS-350(1) LPS-350
Special Power Supply
Integrated Xenon
Integrated Xenon
Integrated Xenon
Variable Attenuator(2) VAA-220A(2) VAA-220A(2) VAA-220A
Monitor Detector(s)
SD-S1
SD-S1
SD-S1
Detector Filters (in Filter Holder)
Photopic
Photopic
Photopic
System Software
HELIOSense
HELIOSense
HELIOSense
Spectral Radiance Monitor (Type, Spectral Range)
CDS-800
CDS-800
CDS-800
STANDARD SYSTEM CALIBRATIONS (NIST Traceable)			
Luminance
Yes
Yes
Correlated Color Temp (All lamps matched & w/VA position)
Yes
Yes
QTH Only Spectral Radiance (350-2400nm)
Yes
Yes
Xenon Only Spectral Radiance (350-2400nm)
Yes
Yes
QTH & Xenon Spectral Radiance (350-2400nm) & CDS
Yes
Yes
Exit Port Spatial Uniformity
Yes
Yes
Exit Port Angular Uniformity
Yes
Yes
Operational Duration of Calibration (Xe/QTH)
25 hrs / 50 hrs
25 hrs / 50 hrs

www.labsphere.com | sales@labsphere.com

Yes
Yes
Yes
Yes
Yes
Yes
Yes
25 hrs / 50 hrs© 2022 Labsphere, Inc. All Rights Reserved PB-14071-000 Rev 11

A Family: Air Mass (Albedo) Sunlight Source 					Model Number
Smart Part Number

USLR-A12F-XDN2-P
USLR-A12L-XDN2-P
USLR-A12L-UDN1-P
A3XD-N2NN-NNAR-NS00-0000-P A4XD-N2NN-NNAR-NS00-0000-P A4UD-N1NN-NNAR-NS00-0000-P

OPTICAL PERFORMANCE SPECIFICATIONS			UNIFORMITY (EACH LAMP OR COMBINED)			
Spatial Luminance Uniformity over Exit Port (f/4) - All Lamps On+/-1.0%+/-1.0%Angular Uniform FOV (Full Angle) - Degrees / F# / NA - All Lamps On+/-2.0% - 35° / 0.85 / 0.6+/-2.0% - 35° / 0.85 / 0.6+/-1.0%+/-2.0% - 35° / 0.85 / 0.6

QTH ONLY			
Expected Luminance Output: cd/m2
11,50015,50015,500
Expected Illuminance at Port: lux
36,00049,00049,000Est. Peak Radiance: W/m2-sr-um @ 0.95 um
350
460
460
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.95um
1.50E+21
2.20E+21
2.20E+21
Minimum Resolution: lux
3.27E+00
3.27E+00
3.27E+00
Approximate Correlated Color Temperature (QTH)
3000K +/-50K
3000K +/-50K
3000K +/-50K
Typical Lamp Lifetimes (hrs)>500hrs>500hrs>500hrs
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-10% & +/-200K-10% & +/-200K-10% & +/-200K
Est. Output Degradation over 50hrs (% & CCT Shift)-1.0% & +/-20K-1.0% & +/-20K-1.0% & +/-20K
PLASMA OR XENON ONLY			
Expected Luminance Output: cd/m2
30,50042,000
Expected Illuminance at Port: lux
48,000132,000
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
420
550
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl or 0.47 Xe 1.20E+21
1.60E+21
Minimum Resolution: lux
4.36E+00
1.20E+01
Approximate Correlated Color Temperature
6000K +200K/-400K
6000K +200K/-400K
Plasma Rest Mode/Xenon Flicker (Peak-Peak/RMS/Hz) - Typical
4%/1%/5kHz
4%/1%/5kHz
Plasma Test Mode (Peak-Peak/RMS/Hz) - 30 minutes (Rest 5min)
N/A
N/A
Installed Cold Mirror (330-750nm)
Yes
Yes
Typical Lamp Lifetimes (hrs) Plasma/Xenon
400
400
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-50% & -600K-50% & -600K
Est. Output Degradation over 50hrs (% & CCT Shift)-2.5% / -30K
-2.5% / -30K

16,50051,800430
7.16E+20
4.71E+00
6000 +/-300K
4%/1%/5kHz
N/A
None
400
-50% & -600K-2.5% / -30K

PLASMA/XENON & QTH (Both Full Open VA)			
Expected Luminance Output: cd/m2
42,00058,000
Expected Illuminance at Port: lux
84,000
182,000
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
600
820
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl @ 0.47 Xe 1.43E+21
2.50E+21
Approximate Correlated Color Temperature (Xenon & QTH)
5100K +400K/-200K
5100K +400K/-200K

32,000100,800
620
1.03E+21
5400K +400K/-200K

ATTENUATORS			
Number of Steps in Attenuator Range
1.20E+04
1.20E+04
1.20E+04
Dynamic Range/Bits/dB - Full Range of System (both lamps)
2.95E+04/14/89
3.71E+04/15/91
2.55E+04/14/88
INTEGRATING SPHERE			
Coating / Material
Spectraflect
Spectralon
Spectralon
Sphere Internal Diameter: Inches (Meters)
12 (0.3)
11.5 (0.29)
11.5 (0.29)
Frame Type
12 in Cage
12 in Cage
12 in Cage
Output Port Size: Inches (Meters)
4 (0.1)4 (0.1)4 (0.1)SYSTEM COMPONENTS			
QTH Lamps Internal (# , Wattage)
(0)
(0)
(0)
QTH Lamps External (#, Wattage)(1) 150(1) 150(1) 150
Xenon Lamp & Housing
175W
UV 175W
UV 175W
Plasma Lamp & Housing
None
None
None
Power Supplies (# - Model)(1) LPS-350(1) LPS-350(1) LPS-350
Special Power Supply
Integrated Xenon
Integrated Xenon
Integrated Xenon
Variable Attenuator(2) VAD-012(2) VAD-012(2) VAD-012
Monitor Detector(s)
SD-S1
SD-S1
SD-S1
Detector Filters (in Filter Holder)
Photopic
Photopic
Photopic
System Software
HELIOSense
HELIOSense
HELIOSense
Spectral Radiance Monitor (Type, Spectral Range)
CDS-800
CDS-800
CDS-800
STANDARD SYSTEM CALIBRATIONS (NIST Traceable)			
Luminance
Yes
Yes
Correlated Color Temp (All lamps matched & w/VA position)
Yes
Yes
QTH Only Spectral Radiance (350-2400nm)
Yes
Yes
Xenon Only Spectral Radiance (350-2400nm)
Yes
Yes
QTH & Xenon Spectral Radiance (350-2400nm) & CDS
Yes
Yes
Exit Port Spatial Uniformity
Yes
Yes
Exit Port Angular Uniformity
Yes
Yes
Operational Duration of Calibration (Xe/QTH)
25 hrs / 50 hrs
25 hrs / 50 hrs

www.labsphere.com | sales@labsphere.com

Yes
Yes
Yes
Yes
Yes
Yes
Yes
25 hrs / 50 hrs© 2022 Labsphere, Inc. All Rights Reserved PB-14071-000 Rev 11

A Family: Air Mass (Albedo) Sunlight Source						Model Number
Smart Part Number

USLR-A12F-XMN2-P
USLR-A12L-XMN2-P
USLR-A12L-UMN1-P
A3XM-N2NN-NNAR-NS00-0000-P A4XM-N2NN-NNAR-NS00-0000-P A4UM-N1NN-NNAR-NS00-0000-P

OPTICAL PERFORMANCE SPECIFICATIONSUNIFORMITY (EACH LAMP OR COMBINED)				
Spatial Luminance Uniformity over Exit Port (f/4) - All Lamps On+/-1.0%+/-1.0%+/-1.0%Angular Uniform FOV (Full Angle) - Degrees / F# / NA - All Lamps On+/-2.0% - 35° / 0.85 / 0.6+/-2.0% - 35° / 0.85 / 0.6+/-2.0% - 35° / 0.85 / 0.6
QTH ONLY			
Expected Luminance Output: cd/m2
11,50015,500
Expected Illuminance at Port: lux
36,00049,000Est. Peak Radiance: W/m2-sr-um @ 0.95 um
350
460
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.95um
1.50E+21
2.20E+21
Minimum Resolution: lux
7.20E-01
9.80E+01
Approximate Correlated Color Temperature (QTH)
3000K +/-50K
3000K +/-50K
Typical Lamp Lifetimes (hrs)>500hrs>500hrs
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-10% & +/-200K-10% & +/-200K
Est. Output Degradation over 50hrs (% & CCT Shift)-1.0% & +/-20K-1.0% & +/-20K

15,50049,000460
2.20E+21
9.80E+01
3000K +/-50K>500hrs-10% & +/-200K-1.0% & +/-20K

PLASMA OR XENON ONLY			
Expected Luminance Output: cd/m2
30,50042,000
Expected Illuminance at Port: lux
42,000132,000
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
420
550
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl or 0.47 Xe 1.20E+21
1.60E+21
Minimum Resolution: lux
8.40E+01
2.64E+00
Approximate Correlated Color Temperature
6000K +200K/-400K
6000K +200K/-400K
Plasma Rest Mode/Xenon Flicker (Peak-Peak/RMS/Hz) - Typical
4%/1%/5kHz
4%/1%/5kHz
Plasma Test Mode (Peak-Peak/RMS/Hz) - 30 minutes (Rest 5min)
N/A
N/A
Installed Cold Mirror (330-750nm)
Yes
Yes
Typical Lamp Lifetimes (hrs) Plasma/Xenon
400
400
Est. Lamp Degradation Over Lifetime (% & CCT Shift)-50% & -600K-50% & -600K
Est. Output Degradation over 50hrs (% & CCT Shift)-2.5% / -30K
-2.5% / -30K

16,50051,800430
7.16E+20
1.04E+00
6000 +/-300K
4%/1%/5kHz
N/A
None
400
-50% & -600K-2.5% / -30K

PLASMA/XENON & QTH (Both Full Open VA)			
Expected Luminance Output: cd/m2
42,00058,000
Expected Illuminance at Port: lux
78,000
182,000
Est. Peak Radiance: W/m2-sr-um @ 0.60um Pl @ 0.47 Xe
600
820
Est. Peak Irradiance @ Port: Photons/s-m2-um @ 0.60um Pl @ 0.47 Xe 1.43E+21
2.50E+21
Approximate Correlated Color Temperature (Xenon & QTH)
5100K +400K/-200K
5100K +400K/-200K

32,000100,800
620
1.03E+21
5400K +400K/-200K

ATTENUATORS			
Number of Steps in Attenuator Range
1.00E+04
1.00E+04
1.00E+04
Dynamic Range/Bits/dB - Full Range of System (both lamps)
1.34E+05/17/102
1.48E+05/17/102
1.02E+05/16/99
INTEGRATING SPHERE			
Coating / Material
Spectraflect
Spectralon
Sphere Internal Diameter: Inches (Meters)
12 (0.3)
11.5 (0.29)
Frame Type
12 in Cage
12 in Cage
Output Port Size: Inches (Meters)
4 (0.1)4 (0.1)

Spectralon
11.5 (0.29)
12 in Cage
4 (0.1)SYSTEM COMPONENTS			
QTH Lamps Internal (# , Wattage)(0)(0)
QTH Lamps External (#, Wattage)(1) 150(1) 150
Xenon Lamp & Housing
175W
175W
Plasma Lamp & Housing
None
None
Power Supplies (# - Model)(1) LPS-350(1) LPS-350
Special Power Supply
Integrated Xenon
Integrated Xenon
Variable Attenuator(2) VA-MM(2) VA-MM
Monitor Detector(s)
SD-S1
SD-S1
Detector Filters (in Filter Holder)
Photopic
Photopic
System Software
HELIOSense
HELIOSense
Spectral Radiance Monitor (Type, Spectral Range)
CDS-800
CDS-800(0)(1) 150
UV 175W
None(1) LPS-350
Integrated Xenon(2) VA-MM
SD-S1
Photopic
HELIOSense
CDS-800

STANDARD SYSTEM CALIBRATIONS (NIST Traceable)			
Luminance
Yes
Yes
Correlated Color Temp (All lamps matched & w/VA position)
Yes
Yes
QTH Only Spectral Radiance (350-2400nm)
Yes
Yes
Xenon Only Spectral Radiance (350-2400nm)
Yes
Yes
QTH & Xenon Spectral Radiance (350-2400nm) & CDS
Yes
Yes
Exit Port Spatial Uniformity
Yes
Yes
Exit Port Angular Uniformity
Yes
Yes
Operational Duration of Calibration (Xe/QTH)
25 hrs / 50 hrs
25 hrs / 50 hrs

Yes
Yes
Yes
Yes
Yes
Yes
Yes
25 hrs / 50 hrs

www.labsphere.com | sales@labsphere.com© 2022 Labsphere, Inc. HELIOS is a Registered Trademark© 2022of Labsphere, Inc. All Rights Reserved PB-14071-000 Rev 11