Infinity XE 5011-5041

Single nitrogen source for multiple applications
Part Number : See Reverse

Service Kit : See Reverse

Your local gas generation partner

Description
Producing between 10 and 520 L/min of high purity nitrogengas, Infinity XE 50 series gas generators can comfortably supply
multiple laboratory instruments. Offering variable N2 purities(95- 99.5%) with a built in purity fail-safe, you can now effectively
meet the demands of numerous applications such as ELSD, NMR& Glovebox, as well as LC-MS. With no moving parts or mechanicalsound, our Infinity range is easy to service and requires minimalmaintenance.
Peak Scientific's Infinity XE series generators have been engineered
to provide nitrogen to laboratories that utilize an external sourceof compressed air, which meets a minimum quality grade of ISO
8573-1:2010 Class 1.4.1. Where required, Peak can provide assistanceon external compressors & pre/post filtration as part of delivering acomplete solution for your needs.

Applications)

LC-MS)

NMR)

ELSD)

FT/MS)

Sample Evaporators)

Glovebox

500.00mm

Key Features

500.00mm

995.00mm)

Flow rates ranging from 10-520 L/min*)

Variable N2 purities, ranging from 95-99.5%*)

Maximum output pressure of 135 psi)

Built upon tried & tested membrane technology)

Compact, space saving design)

Purity safeguard, warning & auto-shutdown)

Energy efficient eco-mode ensures low running cost
* Dependent on inlet pressure, inlet/outlet flow & inlet air quality

INFINITY XE 5011-5041 Dimensions

Copyright 2022 Peak Scientific - Rev - Infinity XE 5011-5041 Data sheet  Rev 7 RSID 1974  EN - 01/02/22

Technical Specifications

5011

502150315041

Flow Rate Min - Max (L/min)*10-13020-26030-39040-520Purity N2 Min - Max *95%-99.5%

Max Outlet Pressure

135 psi

Inlet Air Quality Requirement

ISO8573-1

Inlet Air Pressure Requirement60-145 psi (depending on outlet pressure requirements)

Voltage100-230 VAC +/- 10%Frequency

50/60 HzCurrent

0.4A @ 100V / 0.174A @ 230V

Input ConnectionWC20 Plug

Power Cord (Supplied)

C19 Socket to local connection

Power Consumption

40 Watts

Operating Temperature

5°C - 30°C / 41°F - 86°F

Pollution Degree

2Installation Category

IIDimensions mm (inches) W x D x H

500 (19.7) x 500 (19.7) x 995 (39.2)

Weight kg (lbs)

75kg (165lbs)

80kg (176lbs)

85kg (187lbs)

90kg (198lbs)

Shipping weight kg (lbs)

91kg (200lbs)

96kg (211lbs)

101kg (222lbs)

106kg (233lbs)

33001903300191

3300192*Dependant on inlet pressure, inlet/outlet flow and Inlet Air Quality

Ordering Information
Part Number

3300189

Annual Service Kit
Fixed Price Preventative Maintenance

Visit: www.peakscientific.com/ordering

Complete/Standard Maintenance Plan

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2022 Peak Scientific - Rev - Infinity XE 5011-5041 Data sheet  Rev 7 RSID 1974  EN - 01/02/22