Genius NM32LA-110V
Nitrogen Generator for LC-MS
Part Number : 10-6520

Service Kit : 08-4780

Your local gas generation partner

Description
The Genius NM32LA-110V is a variant of our best-selling NM32LA
nitrogen generator for LC-MS, engineered to operate at 110V rather
than 230V as with the standard model. This tailored solution forlaboratories operating at 110V (eg. USA), removes the requirement
for a step-up transformer to boost supply to 230V in order torealize the benefits of the NM32LA.
The culmination of over a decade’s work perfecting an on-site
nitrogen gas generation for LC-MS, the Genius NM32LA is thepinnacle of our Genius series. NM32LA has become a proven andreliable on-demand nitrogen solution in countless laboratories
worldwide. This system’s popularity is underpinned by its
compatibility with most mass specs available in the market.

Applications)

Perfect for labs with 110V
electrical supply)

600.00 mm

Ideal nitrogen gas sourcefor MS instruments
designed to operate in
110V facilities (eg. Waters
Acquity QDa)

713.00 mm
750.00 mm

713.00 mm

Genius Nm32la-110v Dimensions

Copyright © Peak Scientific Instruments Ltd 2021 — Genius NM32LA-110V Rev 8 RSID 460 EN —16/06/2021

Key Features)

Compressor based solution, generator can run on its own
without the need for an external source of air)Proven and reliable with 1000’s in the field worldwide)

110V electrical supply compatible, no step-up
transformer required)Duty and service indicators)

Latest generation, high specification compressor located ininsulated chamber to minimize noise and vibration)

Once connected to the instrument and power is supplied, thesystem is fully operational)

The most cost effective method of gas supply)

Remove the hassle and safety concerns of ordering andchanging out pressurized cylinders)

Gas is supplied on demand so generator works to your scheduleTechnical Specifications

Genius NM32LA-110V

Max Flow Rate

32 L/minMax Pressure

100 psi / 6.9 bar

Max Relative Humidity

80% Non-Condensing

Max Altitude2000 Metres

Particles< 0.01µm

Gas Outlets

1x 1/4” BSP Female

Phthalates

None

Suspended Liquids

None

Operating Temperature

5ºC - 35ºC / 41ºF - 95ºF

Electrical Requirements

100-115V 60Hz 11A

Power Consumption

1265 Watts

Noise Level

59 dB

Generator Dimensions (HxWxD)
Generator Weight713 x 600 x 750 mm/28.1 x 23.6 x 29.5 in
95 Kg (209lbs)

Ordering Information
Part Number

10-6520

Annual Service08-4780

Standard Maintenance Plan

09-3110Complete Maintenance Plan

09-3010Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2015

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com
Copyright © Peak Scientific Instruments Ltd 2021 — Genius NM32LA-110V Rev 8 RSID 460 EN —16/06/2021