Page 42

LOW HEIGHT COLUMN 2029/2030
SERIES FORCE TRANSDUCERSFEATURES• Sealed for Environmental Protection• Water Immersion to 300 ft.
• Integral Cable• External Male Threads• 5pt Tension & Compression Calibration• Proprietary Modulus Compensated Strain Gages• Optional Female Thread• Metric Capacities Available- Consult Factory

SPECIFICATIONS
ModelPARAMETERS

Capacity

2029

2030

U.S. lbf. 500, 1000, 5K 2.5K, 5K, 7.5K

ACCURACY

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated OutputCreep, % in 20 minutes
Zero Balance, % Rated Output+/-0.250+/-0.250+/-0.150+/-0.050+/-0.030+/-1.0+/-0.250+/-0.250+/-0.150+/-0.050+/-0.030+/-1.0Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ °F
Effect on Zero, % Rated Output/ °C+15 to +115
-10 to +45-65 to +200-55 to +90+/-0.0015+/-0.0027+/-0.0015+/-0.0027+15 to +115
-10 to +45-65 to +200
-55 to +90+/-0.0015+/-0.0027+/-0.0015+/-0.0027

Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm

1
1
15
15
1000 +100/-10 1000 +100/-101000 +100/-10 1000 +100/-105000 (@ 50VDC)

Safe Overload Range, % Rated Capacity
Weight lb
Flexure Material+/-150+/-1500.7
0.8
Stainless Steel Stainless SteelTEMPERATURE

ELECTRICALMECHANICALPage 432121 SERIES COMPACT STAINLESS STEELCOMPRESSION LOAD CELLSFEATURES• Compact Size
• Accuracy-.05% - .07% of Rated Output• Optional Full Encapsulation for Harsh Environments• 5pt Compression Calibration• Proprietary Modulus Compensated Strain Gages• Optional Connectors or Integral Cable
Note: shown with Optional Female Threads.

DIMENSIONS
Model

2029

500, 1000,
Capacity
5K lbfinch
Dimensions
1.
3.50
1.70
2.
R 1.573.
5/8-18
4.
.75
5.

2030

2.5K, 5K,
7.5K lbf
inch
3.50
1.70
R 1.57
3/4-16.75

SPECIFICATIONS
Model

PARAMETERS

ACCURACYCapacity

U.S. lbf.
Metric kN+/-0.070+/-0.070+/-0.050+/-0.010+/-0.025+/-1.0Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ °F
Effect on Zero, % Rated Output/ °C+15 to +115
-10 to +45-30 to +200
-35 to +93+/-0.0020+/-0.0036+/-0.0020+/-0.0036

Rated Output, mV/V, Nominal
Excitation Voltage, VDC MaximumInput Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm
Bayonet Connector

215350350
5000
MS3102E-14S-6P

Safe Overload Range, % Rated Capacity
Weight lb
Weight kg
Flexure Material+/-150
3.6
1.6
Stainless Steel

ELECTRICALMECHANICAL© 2015 Tovey Engineering, Inc.

10K, 25K lbf
45, 100

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

TEMPERATURE

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com

2121

DIMENSIONS
Model

Capacity
Dimensions
1.
2.
3.
4.
5.
6.
7.
8.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

9.

2121

2121

10K, 25K lbf 45, 100 kN
inch

3.25
2.69
2.88
1.25
4.40
1.722.04
1/2-20.5 Deep

mm

82.55
68.3
73.15
31.75
101.6
10.16
43.69
51.82
M12 X 1.25
12.7 Deep