3130

Portable PressureCalibrator

Technical DataFeatures• Measure and generate pressures from vacuum to 2 MPa(300 psi, 20 bar)
• Internal pump can generate vacuum to -80 kPa (-12 psi,-0.8 bar) or pressure to 2 MPa (300 psi, 20 bar)
• Supply pressure connection allowing the use of external gas
supply up to 2 MPa (300 psi, 20 bar)
• Includes variable volume for fine adjustment of pressures• Pressure measurement accuracy of ±(0.025 % ofreading + 0.01 % FS)• Electrical measurement and 24 volt supply forclose looped calibrations• Measure or generate 4 to 20 mA• Measure 0 to 30 V dc
• Powered by internal, rechargeable, highcapacity NiMH battery or universal ac mains
adapter• Compatible with Fluke 700P and 750P Pressure
Modules

The 3130 Portable Pneumatic Pressure Calibrator isideal for calibrating pressure transmitters, transducers, gauges and similar devices. The 3130 contains
everything you need to generate, control andmeasure pressure, as well as read the output of thedevice under test (DUT).

Pressure generation and control
The internal, motorized pump provides an easy,
efficient alternative to time consuming, tiring handpumps. For applications that involve filling a large
volume with pressure, the 3130 allows for connection to an external gas supply such as compressed
plant air. The pressure can be fine-tuned using thevariable volume.

Pressure measurement
The 3130 features an onboard pressure sensor
with a full scale of 2 MPa (300 psi, 20 bar) andan accuracy of ± 0.025 % reading + 0.01 % FS
(includes precision, one-year stability, uncertainty of
the calibration standard, and temperature effects). Inaddition, the 3130 can be used with the Fluke 700P
or 750P pressure module series to improve measurement performance across the entire range.

Electrical measurement
The 3130 offers electrical measurement capabilities
for calibrating pressure transducers and transmitters,
including measurements of 4 to 20 mA or 0 to 30 V
dc. In addition, the 3130 provides 24 V dc supply topower the DUT and can generate current in the 4 to20 mA range.

Portability
The 3130 is built into a ruggedized case with internal storage for power supply,
test leads, and fittings. In addition, it includes a rechargeable, NiMH battery,allowing for approximately 50 hours of operation. When using the internal
pump, the battery is sufficient to provide up to 100 calibration cycles to 300 psi.
The battery is recharged by simply plugging the unit in using the included
universal mains adapter.

Specifications
Ordering Information
Models3130-G2M

Portable Pressure Calibrator
with US, EUR, UK, and China/
Australia line cords (standardcalibration)
3130-G2M/C
Portable Pressure Calibrator
with US, EUR, UK, and China/
Australia line cords (ISO 17025
accredited calibration)
3130-G2M-2
Portable Pressure Calibrator
with Brazil, Italy, Switzerland,and India line cords (standardcalibration)
3130-G2M-2/C Portable Pressure Calibrator
with Brazil, Italy, Switzerland,
and India line cords (ISO 17025
accredited calibration)

Environmental
Operating temperature
Storage temperaturePower requirements
Battery
Operating time
On full charge
Using internal pumpPhysicalDimensions
Weight
EMI/RFI conformance
Connectors/ports
Included accessories-10 °C to +50 °C-20 °C to +60 °C
12 V dc (Universal ac adapter/charger supplied)Internal 3800 mAh advanced NiMH pack
Approx 50 hours (Measure only or external air; nopump)Approx 100 calibration cycles to 300 psi
15.25 in L x 12 in W x 7 in D
~7 kg (15 lb)
EN61326:2006 Annex A
1/8 in NPT (External supply port and test port)
Manual, NIST-traceable certificate, test leads, universal ac adapter/charger

RangesPressure (internal pump)Pressure (external air)mA
VoltsEngineering units-80 kPa to 2 MPa (-12 to 300 psi, -0.8 to 20 bar)
0 to 2 MPa (0 to 300 psi, 0 to 20 bar)
0 to 24.000 mA
0 to 30.000 V dc
psi, bar, mbar, kPa, MPa, kgf/cm2, mmH2O @ 4 °C,mmH2O @ 20 °C, cmH2O @ 4 °C, cmH2O @ 20 °C,inH2O @ 4 °C, inH2O @ 20 °C, inH2O @ 60 °F, mmHg
@ 0 °C, inHg @ 0 °C
Instrumental measurement uncertainty
Pressure±(0.025 % of reading + 0.01 % FS)
mA±(0.015 % of reading + 0.002 mA)
Volts±(0.015 % of reading + 0.002 V)
Temperature effect (all functions)
No effect on accuracy on all functions from 15 °C to 35 °C
Add ± 0.002 % F.S./°C for temps outside of 15 °C to 35 °C

Fluke Calibration. Precision, performance, confidence.™

Fluke Calibration
PO Box 9090,Everett, WA 98206 U.S.A.

Fluke Europe B.V.
PO Box 1186, 5602 BD
Eindhoven, The Netherlands

For more information call:In the U.S.A. (877) 355-3225 or Fax (425) 446-5116
In Europe/M-East/Africa +31 (0) 40 2675 200 or Fax +31 (0) 40 2675 222
In Canada (800)-36-FLUKE or Fax (905) 890-6866
From other countries +1 (425) 446-5500 or Fax +1 (425) 446-5116
Web access: http://www.flukecal.com©2014 Fluke Calibration. This information is preliminary and subject to change.
Printed in U.S.A. 12/2015 4265091d-en
Pub_ID: 12033-eng
Modification of this document is not permitted without written permission
from Fluke Calibration.

2 Fluke Calibration

3130 Portable Pressure Calibrator