Page 365200 SERIES UNIVERSALHIGH CAPACITY LOAD CELLS

5200 SERIES UNIVERSALHIGH CAPACITY LOAD CELLSFEATURES• Universal Tension and Compression• 4 Column Cell
• .1% Accuracy• Accomodates High Side and Off-Center Loads• 5pt Tension & Compression Calibration• Proprietary Modulus Compensated Strain Gages• Metric Capacities Available - Consult Factory

SPECIFICATIONS
Model

PARAMETERS

Capacity

U.S. lbf.

DIMENSIONS
Model

5230

Capacity 100K, 135K lbf
Dimensions
inch
8.00
1.
R 3.23
2.
5.00
3.
5.49
4.

5230

5240

5260

5280

100K, 135K

250K500K1000K+/-.100+/-.100+/-.100+/-.010+/-.025+/-1.0+/-.100+/-.100+/-.100+/-.010+/-.025+/-1.0+/-.100+/-.100+/-.100+/-.010+/-.025+/-1.0+/-.150+/-.150+/-.150+/-.010+/-.025+/-1.0+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-30 to +200-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027

5240

52605280

250K lbf
inch

500K lbf
inch

1000K

5.

2-12
2.4 Deep

8.00
R 3.48
5.50
5.99
2-12
2.75 Deep12.00
R 4.96
8.40
8.90
4-8
4.0 Deep

6.

3.30

4.00

6.00

Consult Factory

ACCURACY

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

TEMPERATURE
Range, Compensated, °FRange, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ ° F
Effect on Zero, % Rated Output/ ° C

OPTIONS
Additional bridges
Output adjustment to standard
Compression overload protection
Integral cable
Custom connectors

ELECTRICAL
Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, MegohmBayonet Connector

215700700
5000

221515700700700700
5000
5000
PT02E-10-6P (MS3112E-10-6P)

Connector Protectors

2157007005000

ACCESSORIESMating connector
Loading fixtures

MECHANICAL
Safe Overload Range, % Rated Capacity
Weight lb
Flexure Material+/-15034+/-150
42+/-150
129

Interconnect cable+/-150
331

Shunt calibration resistor
Instrumentation

Steel

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 37

Page 365200 SERIES UNIVERSALHIGH CAPACITY LOAD CELLS

5200 SERIES UNIVERSALHIGH CAPACITY LOAD CELLSFEATURES• Universal Tension and Compression• 4 Column Cell
• .1% Accuracy• Accomodates High Side and Off-Center Loads• 5pt Tension & Compression Calibration• Proprietary Modulus Compensated Strain Gages• Metric Capacities Available - Consult Factory

SPECIFICATIONS
ModelPARAMETERS

Capacity

U.S. lbf.

DIMENSIONS
Model

5230

Capacity 100K, 135K lbf
Dimensions
inch
8.00
1.
R 3.23
2.
5.00
3.
5.49
4.

5230

5240

5260

5280

100K, 135K

250K

500K

1000K

+/-.100+/-.100+/-.100+/-.010+/-.025+/-1.0+/-.100+/-.100+/-.100+/-.010+/-.025+/-1.0+/-.100+/-.100+/-.100+/-.010+/-.025+/-1.0+/-.150+/-.150+/-.150+/-.010+/-.025+/-1.0+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027

5240

5260

5280

250K lbf
inch

500K lbf
inch

1000K

5.

2-12
2.4 Deep

8.00
R 3.485.50
5.99
2-12
2.75 Deep12.00
R 4.96
8.40
8.90
4-8
4.0 Deep

6.

3.30

4.00

6.00

Consult Factory

ACCURACY

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

TEMPERATURE
Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ ° F
Effect on Zero, % Rated Output/ ° C

OPTIONS
Additional bridges
Output adjustment to standard
Compression overload protection
Integral cable
Custom connectors

ELECTRICAL
Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm
Bayonet Connector

215700700
5000

221515700
700
700
700
5000
5000
PT02E-10-6P (MS3112E-10-6P)

Connector Protectors

215700700
5000

ACCESSORIESMating connector
Loading fixtures

MECHANICAL
Safe Overload Range, % Rated Capacity
Weight lb
Flexure Material+/-15034+/-150
42+/-150
129

Interconnect cable+/-150331

Shunt calibration resistor
Instrumentation

Steel

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 37