https://www.kadashika.jp

https://www.kadashika.jp

Preface
Dear users:
Thank you for using Handy Medical’s Digital Dental X-Ray Imaging System
HDR-300/400、HDR-500/600 and the trust to Handy Medical’s products. We will
provide our best to ensure the satisfaction when you using the Digital Dental X-Ray
Imaging System HDR-300/400、HDR-500/600 .
The Digital Dental X-Ray Imaging System HDR-300/400、HDR-500/600 is
operated through the software HandyDentist. HandyDentist is designed to take fulladvantage of HDR-300/400、HDR-500/600 . To get the best imaging andprocessing quality, the combine between HDR-300/400、HDR-500/600 and
HandyDentist is necessary.
To ensure your safety and effective use of the Digital Dental X-Ray Imaging
System HDR-300/400、HDR-500/600 , please read the manual carefully before use.

Handy Medical remind: Any copy of the contents in this manual is not allowed
without the permission of Handy.

https://www.kadashika.jp

Content
1. Notice.......................................................................................................................................1
1.1 Indication of Use..................................................................................................... 1
1.2 Brief Introduction of this Manual........................................................................... 1
1.3 Manufacturer............................................................................................................1
1.4.Packing.....................................................................................................................1
1.5 Symbols....................................................................................................................1
2. Safety issue............................................................................................................................ 2
2.1 Check Sensor and controller before using them................................................ 2
2.2 Protect Sensor from Potential ESD Damage...................................................... 2
2.3 Do Not Touch Exposed Connectors on Non-Medical Equipment and thePatient at the Same Time............................................................................................. 2
2.4 Ensure Proper System and PC Workstation Installation and Operation......... 2
2.5 Safety Classifications............................................................................................. 2
2.6 Conditions required in operation, transportation, and storage..........................2
3. Waste Electrical and Electronic Equipment........................................................................3
3.1 Background..............................................................................................................3
3.2 WEEE Marking........................................................................................................ 3
3.3 Reporting..................................................................................................................3
3.4 WEEE from Users other than Private Households.............................................3
3.5 Information for Reuse Centers, Treatment and Recycling Facilities................3
3.6 Warning and Safety Instructions........................................................................... 3
3.7 Hygiene and Disinfection Instruction....................................................................4
4. Handy Sensor General Introduction....................................................................................4
4.1 Functional Components......................................................................................... 4
4.2 Technical Specifications......................................................................................... 5
4.3 Sharing the Sensor Between Rooms................................................................... 5https://www.kadashika.jp

4.4 Using the different Positioning Systems.............................................................. 5
4.5 Attentions................................................................................................................. 5
5. Imaging Software General Introduction.............................................................................. 6
5.1 Computer System Requirements..........................................................................6
5.2 Imaging Software.................................................................................................... 6
6. Working Sketch...................................................................................................................... 6
7. Driver Installation................................................................................................................... 7
8. Software Installation.............................................................................................................. 8
9. Register Key......................................................................................................................... 10
10 Install Calibration File......................................................................................................... 11
11 Software Operation............................................................................................................. 11
12 Acquiring an Image.............................................................................................................20
12.1 Preparing the HDR sensor................................................................................ 20
12.2 Preparing for Acquisition....................................................................................20
12.3 Launching the X-Ray..........................................................................................21
12.4 How to get X-Ray Image................................................................................... 21
13. Disinfection and cleaning..................................................................................................24
14. Maintenance.......................................................................................................................24
14.1. Visual Inspection................................................................................................24
14.2. Periodic Maintenance....................................................................................... 24
14.3. Cable Care......................................................................................................... 25
14.4. Damaged or Non-Functioning Sensor............................................................ 25
15. Warranty............................................................................................................................. 25
16.Adjust exposure x-ray dose. ............................................................................................25
17.Diagnose Error Message...................................................................................................26
Annex A. EMC table.
中文说明书 在 本说明书第31 页https://www.kadashika.jp

1. Notice
1.1 Indication of Use
The Digital dental X-ray imaging system is intended to generate intraoral x-ray image ,require toworking with x-ray source and imaging software in dental clinic for dentist and orthodontists.
1.2 Brief Introduction of this ManualThis manual consist of the safety issue, HDR-500 and HDR-600 brief introduction, software
introduction, how to use sensor and the warranty policy.
1.3 Manufacturer
Manual Name: Digital Dental X-Ray Imaging System Manual
Revision Number: 01
Print Date: Jan, 2015
The Brand names and logos reproduced in this manual are copyright.
Handy is a trademark of Handy used under license.
The system shall be in accordance with IEC60601-1. The person connect the product to the hostcomputer shall insure its compliance.
0123
Manufacturer
Shanghai Handy Medical Equipment Co., Ltd
2nd Floor, Blg 3, No.2688 Jinqiu Rd., Baoshan District, Shanghai 600444 China
European Representative:
Shanghai International Holding Corp.GmBH
1.4.Packing
1. Digital Dental X-Ray Imaging System
2. CD
3. Manual
4. Holder
5. Warranty and Quality Control Card
1.5 Symbols
Marking and Labeling Symbols

1pc
1pc
1pc
1pc
1 pc

1

https://www.kadashika.jp

Marking and Labeling Symbols
Label Location
The following Figure indicates the label locations of The HDR-500 and HDR-600 Systems.
Figure 1 HDR-500 and HDR-600 Label Locations

2. Safety issue
2.1 Check Sensor and controller before using them
Before each usage, check the outer surface of the Sensor and controller for any signs of physicaldamage or defect. Sensor and controller surfaces should have a smooth finish, with no evidenceof chipping or damage. If detected, contact your local distributor of this product for further
instructions.
2.2 Protect Sensor from Potential ESD Damage
Like other electronic devices, Sensor is susceptible to electrostatic discharge (ESD), particularly
when the device is used in or around carpeted areas or low humidity environments. During cablereplacement, when Sensor contacts are exposed, it is especially important to protect the device
from potential ESD damage. Touching a metal surface prior to replacing the cable will reduce
the risk of damaging Sensor components by accidental static discharge. The use of anti-static
floor mats or floor treatments (for example Staticide 6005/6002) will also help eliminate staticbuild-up in your office.
2.3 Do Not Touch Exposed Connectors on Non-Medical Equipment and the Patient at theSame Time
When the Sensor and controller are in use, avoid touching exposed connectors on non-medical
electrical equipment and the patient at the same time. The human body is capable of conducting
electrical current and may cause a shock hazard to patients if appropriate safety practices are notobserved.
2.4 Ensure Proper System and PC Workstation Installation and Operation
The Sensor and controller have been determined to be in accordance with international safety
standards and are deemed suitable for use within the patient area, which extends from the patient
for a distance of 5 ft (1.5m). To comply with these standards, do not operate non-medical
equipment (such as a PC workstation) inside the patient area. Outside the patient area, thepresence of approved non-medical grade equipment and Listed / Approved / IEC 60950-1
certified Information Technology
Equipment (ITE) computer equipment is acceptable. The host computer (PC workstation) shouldbe CE-approved and conform to the Low
Voltage [73/23/EC] and EMC Directive [89/336/ERC]. The system shall be in accordance with
IEC60601-1-1. The person connect the productto the host computer shall insure its compliance. Also, to help ensure optimal performance,
ensure that all software programs residing on the workstation are virus-free and have beenadequately tested so they will not impact imaging applications after installation. Any questions
please contact your local distributor
2.5 Safety Classifications
Safety Type: Class-Ⅱ BF
Power: DC5V Maximum 200MA.Rate of Work: Max 1W
Degree of protection against water resistance: IPX7 (Only applicable for the sensor head part.)
Non AP equipment, non APG equipment
Mode of operation: Continuous operation
2.6 Conditions required in operation, transportation, and storage
2.6.1 Operating conditions
Environment Temperature : +10℃～+40℃

2

https://www.kadashika.jp

Environment relative humidity : ≤95%；
Air pressure : 860hPa～1060hPa；
2.6.2 Transport and Storage conditions
Environment Temperature: -25℃~+60℃;
Environment relative humidity : 10%~93%;
Air pressure : 860hPa~1060hPa;
3. Waste Electrical and Electronic Equipment
3.1 Background
The European Union’s Waste Electrical and Electronic Equipment (WEEE) Directive(6002/96/EC) has been implemented in member states as of August 13, 6005. This directive,
which seeks to reduce the waste of electrical and electronic equipment through re-use, recycling,
and recovery, imposes several requirements on producers. Handy Medical and its Dealers arecommitted to complying with the Directive.
3.2 WEEE Marking
All Handy products subject to the WEEE Directive and shipped after August 13, 6005 will becompliant with the WEEE marking requirements. These products will be identified with the“crossed-out wheeled bin” WEEE symbol shown below, as defined in European Standard EN
50419, and in accordance with WEEE Directive 6002/96/EC.

This “crossed-out wheeled bin” symbol on the product or on its
packaging indicates that this product must not be disposed of with otherunsorted municipal waste. Instead, it is user’s responsibility to dispose of EE waste equipmentby handing it over to a designated collection point for the reuse or recycling of waste electricaland electronic equipment. The separate collection and reuse or recycling of Electrical &Electronic waste equipment will help to conserve natural resources and ensure that it is recycled
in a manner that protects the environment and human health. For more information about where
you can drop off your waste equipment for recycling, please contact your local officials.
3.3 Reporting
According to the WEEE Directive, Handy Medical or its Dealers will ensure that informationneeded to calculate the financial obligations with respect to EEE products will be provided as
required.
3.4 WEEE from Users other than Private Households
According to the WEEE Directive, Handy Medical or its Dealers will fulfill its obligations forthe management of WEEE from users other than private households.
Furthermore, as required by the WEEE Directive, in order to enable the date upon which theequipment was put on the market to be determined unequivocally, a mark on the equipment will
be placed to specify that the equipment was put on the market after August 13, 6005.
3.5 Information for Reuse Centers, Treatment and Recycling Facilities
After August 13, 6005, and as required by the WEEE Directive, Handy Medical or its Dealers
will provide reuse, treatment, and recycling information for each type of new EEE put on themarket within one year of the date in which the equipment is put on the market.
Information will include the different EEE components and materials as well as the location ofsubstances in these items. The information will be provided as a printed document or inelectronic media (on CD-ROM or by web download, for example)
3.6 Warning and Safety Instructions
For Device:Read and comprehend this Safety Instruction before using the HDR500/600 Systems.

The operation and maintenance of this device must be taken charge by you.This device
only can be operated by the legally qualified persons.If necessary, have a authorized
qualified technician carry out inspection and maintenance operations.

3

https://www.kadashika.jp

This device must be installed in a X-ray room that complies with current installation
standards. From this location, Any visual or audio communication with the patient must
be maintained by you and the Acquisition interface module during exposure.
X-ray equipment is hazardous to patients and the operator if you do not comply with the
exposure safety factors and operating instructions.
This device must not be allowed to be operated if there is the threat of an earthquake.
After an earthquake, ensure that the device is operating satisfactorily before using it
again. Failure to observe patients to hazards.
DO NOT place any objects within the field of operation of the device.
Connect this equipment ONLY to a mains power supply with protective ground to avoid
any risk of electric shock.
Disposing of the device or its components must be executed by a qualified service
technician.
Never be allowed to modify the device.
This device is never allowed to be applied in conjunction with oxygen-richenvironments. Nor intended for apply with flammable anesthetics or flammableagents.Using accessories other than those specified in this document with the exceptionof those sold by Handy Health may result in a lower level of security for the entire
system.
For Computer:
DO NOT place the computer and the peripheral equipment connected toit in theimmediate vicinity of the patient in the unit. Leave at least 1.5m distance between thepatient and the unit. The computer and the peripheral equipment must conform to the
IEC60950 standard.
Read your computer installation guide for details of the data processing system andscreen. Ensure the proper ventilation with leaving a sufficient amount of clear spacearound the CPU.
In order to acquire maximum image quality and visual comfort, direct light reflections
from internal or external lighting should be avoided when position the screen.
3.7 Hygiene and Disinfection Instruction

DO NOT place the sensor in an autoclave environment as which could cause serious
damage to the sensor.
Never immerse the RVG sensor in any solution.
The sensor head should be disinfected after each patient.
Do not apply chemical autoclave for the toothbrush holders and avoid direct contact
with the metallic part of the autoclave.

To prevent from cross-contamination , apply a new hygienic barrier for each new patient
4. Handy Sensor General Introduction
Dental digital x-ray imaging system is consisted of sensor, image controller, image capturesystem and connection cable (USB port) , connected with PC or notebook via USB cable. The
power of controller and sensor is supplied by USB port, require no battery or power charge
system. The whole equipment need to work together with imaging software.
4.1 Functional Components
4.11 HDR sensor


HDR-300/400

HDR-500/600

4

https://www.kadashika.jp

The sensor active surface is flat including the size 1 and size 2.
Size1, universal sensor---Use for regular procedures, both for children and adult .
Size2 sensor---Use for bitewings procedures.
The sensor non-reactive to X-Rays surface, contains the cable attachment.
Figure 2 HDR Sensor

1 Sensor non-reactive to X-Rays surface
2 Sensor active surface
4.2 Technical Specifications
Sensor: APS CMOS sensor
External dimension (mm) :44.3×32.3(HDR-400/600) ; 38.5×27.5(HDR-300/500)
Sensor Active Area(mm) : 36×27(HDR-400/600) ; 30×22.5(HDR-300/500)
Sensor Thickness : 6mm
Dynamic Range : 0~4,096
Power : 5V±0.5V
Image Transfer :USB2.0
Cable Length :.≥ 3m(HDR-500/600) ≥ 2m(HDR-300/400)
4.3 Sharing the Sensor Between Rooms under proper installation and operation
The sensor could be used under net version through HandyServer software and accessed byseveral dentists in different rooms as a working station . In this case, the HandyServer needs to
be installed in one computer which is as the database and other computers in different rooms
could access the data by remote.
Attentions: The HDR sensor and control box conform to the international safety standards andare regarded to be applicable to be used in patient available area, namely, within 1.5m distance.
Within the patient available area, the (computer) devices used should be qualified as a medical
device and is compliant with GB9706.1(IEC 60601-1), YY0505(IEC 60601-1-1) standards;
outside the patient available area, the (computer) devices used other than a medical device
should be compliant with GB4943(IEC 60950); the whole working station configurations should
be compliant with GB9706.15(IEC 60601-1-2) standards and be verified by qualified station
staffs. In order to ensure the device’s normal performances, all programs used in the working
station should be without virus and under verification .
4.4 Using the different Positioning Systems
There are two ways to position the sensor in the patient mouth to get an classic radiology. You
may spend some time to adapt due to the rigidity of the sensor.
One method is angular bisector technique, the other method is paralleling technique. It is just theway to position sensor, which can be chosen by practitioner’s experience.
X-Ray Generator Compatibility
Normally, the sensor is compatible with all generators which meets the present standard of
intraoral radiology. You can use a high frequency or conventional generator. To achieve gaining
better images, the generator must operate with a voltage of 65 to 70kV.
4.5 Attentions
1 As a precision devices, avoid flop,pulling and long-time disinfectant soak
2.Put disposable plastic jackets before used, avoid cross infection and allergy
3.No bite, for fear that the sensor or the jacket been broken
4.Caution for epileptic or psychopath
5.Learn the user manual carefully before use
6.User should be the professional dental or technician

5

https://www.kadashika.jp

5. Imaging Software General Introduction
5.1 Computer System Requirements
Processor: Intel 1.7GHz chip or above;
Memory: Above 2G ;Hard disk: Above 40G;Interface: USB 2.0;
Display: Resolution 1024 × 758 (15") or above
Operating System: Windows 2000/XP/Win7/Win8/Win10（32bit&64bit）
The computer connected to system shall be in accordance with IEC 60950-1:2005.
5.2 Imaging Software
The Handy HDR-300/500 and 400/600 dental imaging system operates with the following
software:
HandyDentist
HandyServer for sharing information between workstations.

HandyDentist Software is a user-friendly working interface that was designed and developed
specifically for radiological diagnosis. It is the common imaging platform for all our digitalsystems for dentistry.
6. Working Sketch
The sensor and control box is connected already, please check the light on control box, before
taking x-ray image. The power indicator on the control box displays yellow light and working
indicator flash green light.
Working Sketch map

HDR-300/400

HDR-500/600

6

https://www.kadashika.jp

7. Driver Installation
1. Insert the CD into CD-ROM or DVD drive, installation screen will appear,

2. Click on Install Driver, or browse CD then double click on the icon
It will show

3. Click, it will show

4. Click, it will show

5. Click, it will show

to continue.

6. Click, the driver is installed. Then connect the sensor to computer via USB cable,
the computer will recognize it, and will show it is ready to use. You can check the drive name in
Device Manager. The right driver name is.

7

https://www.kadashika.jp

8. Software Installation
We provide CD together with camera within the package. To begin the installation, insert
the CD into your CD-ROM drive. The CD is equipped with an Autorun feature and will
start on its own. If the Autorun feature does not work with the CD-ROM drive in yourcomputer,you can launch the installation menu manually: Double-click on the My Computer icon
on your desktop, then open the CD-ROM drive by double clicking.
8.3.1. Click the button
the iconon the interface of the CD or double-clickin the CD, and then it shows:

8.3.2. Click on the button, and then it shows:

8.3.3. The default destination folder is “D:/HandyDentist”. If you want to change theroute, please click on the button of to select the folder
you want.
Then click on, and then it shows

8.3.4. Click on, then it shows:8

https://www.kadashika.jp

8.3.5. There are two selections for creating the icon, please make the selection
according to your needs click

8.3.6. Click on the button

8.3.7. Click on, then it shows:, then it shows:the following window will be shown:

When the installation is finished, the following window will be show:

9

https://www.kadashika.jp

8.3.8. Then click on the button, the whole installation was completed and thesoftware will be launched automatically, and there will be a shortcut
on thedesktop.
9. Register Key
30 days later since the date of software installation, the software will ask for register. Double
click on the target frame, register key windows will come.
Please remember to register the software within 30 days of installation.

Click on Help in menu bar then select Register .

Input the information in the frame, and then click on Send Registration button, in normal
situation, there will be a auto-run email jumping out.
If your computer have no email support server, you may meet the following notice.

Please click on OK and then click on the button of Export Req file .

And then please send this Req file to email : support@handycreate.com
As soon as we receive your req ,we will send back the registration key to you .

After you receive the registration file, please click on Import act file, and click on Active

10

https://www.kadashika.jp

Software button.
10 Install Calibration File
10.1 Install To Handydentist
Provided with every sensor is a disk containing the sensor calibration file. Each calibration fileis unique to the sensor it was shipped with and the file must be installed on every computer
systems before using that sensor. Every calibration file is consisted of 3 files , copy these 3 files
from the CD and then paste into software installed destination folder , the default destination
folder is D:/HandyDentist/Sensor.And then the calibration files installation is finished .
10.2 Install to other software with Twain function. If you prefer to use your own software,pls follow this procedures.
10.2.1 Find the correct installation path.
Connect Handy sensor into the PC and open the software which you are going to use, then enter
into Twain interface, see reference picture below:

Click “Setting”then will show following picture (If the “Setting” button is in gray and can’t beclicked, please unplug the sensor and try again from the start.)

C:\Program Files ” in red mark is the installation path for calibration files in Twain, or,you maychange the path by clicking button in green mark and the new path you set will be the one forcalibration files installation.
10.2.2 Install the Calibration Files
Copy the 3 files under “Calibration file” in CD and paste them into the correct installation
path file folder, then it’s finished.
11 Software Operation
User interface
Double click on the icon
Example Screen:

to launch the software, the HandyDentist window will be shown:

1.System menu of the program windows11

https://www.kadashika.jp

2.Menu bar
3.Tool bar
4.Patient account list
5.Teeth chart
6.Target frame
7. Image information
Menu Bar

Each item contains a drop-down menu, and some of function can be achieved directly from theicon in toolbar
Change language
Click on Tools(T) on the menu bar and select Option.

Click on Display, then select the your target language.

In Recent history part, there are two options, it arranges the order of First name and Last, thisdepends on countries custom .
Please click on OK, it will show

Click OK, the software will be restarted automatically and switch to be the language you
selected.
ToolbarEach button contains a icon and will show a short message to explain corresponding function by
moving the mouse pointer to the icon.
Establish Patient account1.

Click the iconin the toolbar or select new patient in file of menu bar

12https://www.kadashika.jp

Then it will show

Input patient information ,select picture series and you can also input study comments in theframe and then click on OK . The new patient account has been created .

Target frame (18 pcs)
If the first one is full (Maximum 20 images for each account )Click onfor creating another account under the same patient.

For establishing a account for another patient , please select “ new patient ” in this windows

A new patient case has been created successfully.
Delete patient accountPlease select the patient case that you want to delete it will show the following notice window

Click on “Yes”, and this patient will be deleted.
Search a patient case13

https://www.kadashika.jp

Click on this icon, and input the patient information, the target patient patient will be showed.

Input basic information of patient and click on search button to find a detailed patient account.
Please attention that the software only display 30 pcs recent patients case , all others will behiden inside , so it need search tool if the patient case was not display in patient Data area .
Edit basic information of patient
Click on this icon, and the Patient Info window will be show:

If there is something wrong with the basic information of the patient, it could be modified here.
Tile
Pave several pictures together (the maximum quantity is 6).
Keep on pressing the “ctrl” on the keyboard for selecting several pictures at the same time, thenclick on this icon, and the pictures selected will be tiled.

Press “ESC” to exit.
Play
Click on this icon to view the image one by one.
The default time is 3 seconds and it could be changed depend on your requirement.
Press “ESC” to exit.
Edit Image
Select a image then click on this icon to open “ Handy Image” windows, and you can also open
it with double-click on the image. More information refers to 9.3.
Delete Image
Select the image you want to delete then click on this icon. The following windows will show

Choose “yes” or “no” to delete the image or not.

14

https://www.kadashika.jp

File import
Select a picture from your computer.
Export image
Select a captured image then click on this icon to export this image.
Copy & Paste
Right click on the image, there will be a pull-down menu.

Select copy and then choose a empty target frame right click and select paste, this image will bepaste into another frame.
Refresh images
Click this icon to refresh the images, orRight click on any image, and then select Refresh images.
Print Image
Note: To achieve this function, please be sure that the computer has been connected to a printer.
Print one image: left-click on that target frame and then click on print icon in the tool bar
directly.
Print more images: Keep on pressing the “ctrl” on the keyboard and then left click on the target
frames to select several frames at the same time, then click on this icon to print these selected
images.
* Before use the hardware, please chick the equipment icon first:
The iconis for Intraoral Camera or Film Reader.

The iconis for Intraoral X-ray Sensor.

The iconthis software.

is to make other third-party hardwares which have Twain function compatible to

The iconis for Panoramic Machine.

HandyImage(Enhance image)For enhancing the original image quality please double click the target frame with image or
click on the button of

to come into HandyImage.

15

https://www.kadashika.jp

All of these feature buttons are use to change the attributes not the image itself.
All of these tools could be divided into six parts
Color Adjust
Color adjusted by moving the slider either to left or right depending on your needs.
Bright/Contrast
Brightness and contrast could be adjusted also by moving the slider to left the to decrease thebrightness/contrast, move the slider to right to increase.
DesoftTo remove useless soft tissue and adjust gamma value to improve the image quality for thoseunsatisfied pictures:

BeforeAfter
Image process
There are eight feature buttons listed in this section, to achieve its function just by clicking on
corresponding button.

Before contrast

After contrast

To enhance the contrast on the image

Before Sharpen

After sharpen

Inverts the gray shades of the image-negative images appear as positive, and positive images as
negative.

16

https://www.kadashika.jp

Before Negative

After Negative

Click on the colorize button, you’ll notice the significant color contrast between different
elements of the image.Because some differences can be easier to distinguish in color,colorizing
provides antother means to identify potential problems areas during examnations.

Before Colorize

After Colorize

This function can optimize your image.

Before Embossment

After Embossment

Show better structures and improve underexpose or overexpose images’ quality.

Before Normalize

After Normalize

Removes fixed pattern noise from an image.

17

https://www.kadashika.jp

BeforeAfter

Make the images smoother

BeforeAfter

Simulation Diagosis
In order dentist have a better dental practice and more easy to dignose for particular dentition.
Saprodontia,Root Canal, Periodontal, Crown and Diagosis,four different enhanced image makeeasier to check.

And With one button of Dignosis, you will see four different enhanced image.

18

https://www.kadashika.jp

Handy Image Toolbar
To apply the function, click on corresponding button.

Apply this tool for come back to original image at one time .
Undo: Used to withdraw the previous operation
previous operation.

.
Redo: Used to repeat

Choose the display mode you prefer.

This function is used to view the image from different angle .
Click this button,there’s a small window for choosing the magnifier parameter you prefer.

This function was set for making a mark on the image the pinot out the tooth problem and forbetter communication between patient and dentist.

Click on this Drawing icon, then keep on left pressing of mouse, then move the pointer on the
picture, the red line will appear following the mouse pointer .
Click on this icon, the following windows will show

19

https://www.kadashika.jp

Left Click on two points that you want to know the distance. And left frame will show the
distance, and this is only for reference, not accurate.

And if you calibrate the image taken from other brand sensor or intraoral camera, you will needcalibrate first.
By Pressing this button the edited image will be saved automatically. And it will give thefollowing notice.

Click on OK, the edited image will be saved into software automatically.
12 Acquiring an Image
To acquire an image with the HDR sensor, follow the instructions in the presented order.
12.1 Preparing the HDR sensor
To prepare the HDR sensor, follow these steps:
Select an appropriate positioner for the region of interest and the sensor size.
Cover with a disposable hygienic sleeves specifically designed for each sensor size.

Notice: To prevent cross-contamination, use a new hygienic sleeve for each new patient.
12.2 Preparing for Acquisition
To prepare for acquisition, follow these steps:
1 Open software, establishing the patient account2 Clickto access the capture interface,or double click the blank frame.

20

https://www.kadashika.jp

3 Select the x-ray timing according to the region of interest and the patient type (follow the user
instructions of your x-ray generator).
4 Insert the sensor holding it horizontally in the patients mouth. Positioning in the patient’s
mouth depends on region of interest.
5 Approach the x-ray generator tube head to the patient.
6 Align the x-ray tube head with the patient’s tooth and the sensor and make sure that the tube
head is not shaking.
12.3 Launching the X-Ray
To launch the x-ray, follow these steps:
1.
Ask the patient to remain still2.
Position yourself either 2 meters behind the x-ray generator or outside the door.
3.
Keep visual contact with the patient during the x-ray generator.
4.
Trigger the x-ray with the remote control of the x-ray generator.
The image appears in the capture windows of the capture interface.

The image has been captured and the information frame return back to green color. It’s ready tocapture another image, without closing the window.
As soon as the image was captured it will be saved into software.
5.
Check the image quality. If not satisfactory, redo the x-ray.
6.
If satisfactory, remove the generator tube head.
7.
Remove the HDR sensor from patient’s mouth. Remove the hygienic sensor protection.
Notice: DO NOT pull the sensor by its cable when you remove the hygienic protection.
12.4 How to get X-Ray Image
Note: First of all, please confirm that you have installed the calibration files, more informationrefers to chapter 9.
Connect the sensor to the controller then connect it to computer
The power indicator light will shine, and then the working indicator light will flash green light.
Using disposable sheath
Please place the sensor on the holder and cover the sensor with disposable sheath, then place thesensor into patient’s mouth with flat side facing the X-Ray tube.
Check the x-ray exposure settings.
Select one target frame, right click and select Intraoral X-Ray sensor and pls select HDR

21

https://www.kadashika.jp

NEW.

Double click on the target frame again, the capture windows will show:

The capture windows will come and appear flash green light (default setting) .
The sensor ID # of the sensor you are using is shown directly on the title.
And it will remind you if the calibration file hasn’t been installed in the meanwhile.
Active the x-ray source.
After activate the x-ray source , the sensor will detect the x-ray automatically and it will show .

The green flash frame will change to be yellow color , it means that the sensor has detected the
x-ray successfully .And around two seconds later , the image will be captured as below show .

Advanced Settings
If you find out that some images need do some imaging process to enhance the image quality,
you can use Advanced Settings

22

https://www.kadashika.jp

And you can slide these buttons one the left to adjust to what you expect on the right side, andnext time you will take the image just exactly you want to enhance without any other action.
**However, we will recommend you keep the default figures in normal cases.**** To optimize the final image processing quality, please adjust the window level first and saveto a new picture, then using processing function buttons on this saved new picture. **The image has been captured and the information frame return back to green color .It’s ready tocapture another image ,without closing the window .
As soon as the image was captured it will be saved into software .

For taking another x-ray image, please repeat this procedure.
! Important: When the device is not in use, please disconnect it form USB port to avoid
LED indicator be over heated.
! Important: The disposable sheath is for single use only; please change it for every
patient and dispose the used ones under local health authorities guidance on medical
disposable products.
! Important: The disposable sheath used shall conform to ISO10993 biocompatibility
requirements, i.e. no cytotoxic, delayed type hypersensitivity, irritation or endothelial
reaction.

23

https://www.kadashika.jp

Working Sketch map

（1）

（2）

（3）

（4）

Descriptions:(1) x-ray source(2) Tooth(3) Digital x-ray sensor(4) Connect to computer via USB cable(5) Show the image in software
13. Disinfection and cleaning
The product can’t be dipped in liquids, and the disinfectants and cleaning
agents should not be used. The product shall be used with disposable sheet to achieve
the disinfectant purpose (Handy Company do not provider disposable sheets, please
use the disposable sheets which are compliant with ISO10993 biocompatibility
requirements). Periodically clean the outside of the controller and the sensor by soft
cloth with a small quantity of alcohol. The disinfection and cleaning preparations could
damage the CMOS-sensor or the controller. The usage of disinfection and cleaning
preparations is considered to be a breach of the guidelines for intended use. The
manufacturer cannot be held liable for any damage resulting from incorrect usage. The
operator will be held liable and bears all risks.
·Do not put sensor in an autoclave.
• Do not put the sensor in an ultrasonic bath for cleaning and disinfecting.
• Avoid spraying any of the connectors to the CMOS-sensor!
• Remove coarse dirt before disinfection using a soft, lint-free cloth or wipe.
14. Maintenance14.1. Visual Inspection
Like all electrical equipment, the product requires not only correct use, but also visualinspection prior to operation, and routine checks at regular intervals. These precautions
will help ensure that the product operates accurately, safely, and efficiently.
Before operating the system, users shall check it for any signs of physical damage ordefect. If detected, contact your local distributor of this product for further instructions.
14.2. Periodic Maintenance
Periodic maintenance is performed as needed, but at least once a month. It consists ofvarious checks performed by the operator or by a qualified service technician.

24

https://www.kadashika.jp

• Check that the labels are intact, readable, and adhere well to thesurfaces on which
they are positioned• Check that all of the cables are undamaged• Check that there is no external damage to the product which could compromise its
ability to operate safely• Check the installation, then do the step 1, 2 and 5 of operation and check that theindicator lights and indicator area in software are in normal.
14.3. Cable Care
Improper coiling of a sensor's cable is the most common cause of the sensor failure.
The following instruction is important to be followed for preventing cable damage.
• Grasp the connector off the sensor cable when disconnecting the sensor from USB
control box. Pull it gently.
• Once unpacked, never coil the sensor cable, repeated coiling may cause kinks andirreversible damage.
• Store the sensor in its holder when it's not in use.
• Don't let the cable hang on or near the floor where can become tangled.
• Don't tangle the cable during use.
Note: The length is actual only when it was calibrated. If not, the value is just forreference.
14.4. Damaged or Non-Functioning Sensor
In the event of obvious physical damage to the Sensor or in the event that the sensor
can’t work properly, customers shall discontinue use of the Sensor, and contact theirlocal distributor of the products to substitute another Sensor if available.
15. Warranty
We guarantee that the product functions correctly and that there are no faults in thematerial or workmanship for duration of 12 months following the purchase date,according to the following conditions:
Following a reasonable complaint relating to defects or short delivery, we will provide a
replacement or perform repairs. Our factory reserves the right to perform repairs.
Claims of any other nature, damages in particular, are excluded. In case of default andgross negligence or intent, the latter only applies if there are no compelling legal
provisions opposing it.
We shall not be liable for defects or their consequences if they are likely to be a direct
result of actions or modifications by a customer or third party.
The service life of the product is 4 years without unexpected situation.
16.Adjust exposure x-ray dose.
Just like the traditional exposure machine,expose time depend on the machine model,
and the patient exposed tooth part.Following is our recommend for your reference. You
can adjust and set the setting to needs to take more clear images based on your experi
ence.

25

https://www.kadashika.jp

17.Diagnose Error Message
1.Computer can’t recognize the sensor
Try another USB port and reinstall the driver program
2.Controller box does not work(the dedicator light out)
Check the connect interface between controller box and computer
3.Can’t capture picture
Set the x-ray machine. Try plug to the back USB port of computer case , and reinstall thedriver program.
4.No capture program
Close all anti-virus program and reinstall all program
5.Warning “Lack of calibration file”
Pls copy the calibration file from CD to dedicate folder,the default folder is C:\Program
Files\Handydentist\sensor
6.The image taken is too dark or white
Try to reduce or increase x-ray dose and exposure time
7.Can’t use the handydentsist software over a month.
Please export the req to us for the active file to register the software.
8.The image taken has the white vertical light
Did not install calibration file or calibration file is installed in wrong place.

Annex A. EMC table
The following tables provide HDR compliance information to electromagnetic compatibility (EMC) and electromagnetic
immunity (EMI) standards. To ensure conformance, the customer or user must use the HDR in environments that areconsistent with these standards. USB cables used with HDR Interfaces must also comply with the same standards.
Table 1. Guidance and Manufacturer's Declaration - Electromagnetic Emissions
PLEASE NOTE: HDR is intended for use in the electromagnetic environment specified below. The customer or user of HDR
must ensure that it is used in such an environment.

26

https://www.kadashika.jp

Emissions Test

RF emissions
EN 55011

Compliance

HDR uses RF energy only for its internal function.
Therefore, its RF emissions are very low and are not likely
to cause any interference in nearby electronic equipment.

Group 1

RF emissions
EN 55011

Class B

Harmonic emissions
EN 61000-3-2

N/A

Voltagefluctuations/flicker
emissions
EN 61000-3-3

Guidance

HDR is suitable for use in all establishments,including
domestic establishments and those directly connected to the
public low-voltage power supply network that supplies
buildings used for domestic purposes.

N/A

Table 2. Guidance and Manufacturer's Declaration - Electromagnetic Immunity
PLEASE NOTE: HDR is intended for use in the electromagnetic environment specified below. The customer or user of HDR
must ensure that it is used in such an environment.
Immunity Test

EN 60601 Test Level

Compliance level

Electrostatic discharge(ESD)
EN 61000-4-2±6 kV contact±8 kV air

±6 kV contact±8 kV airFloors should be wood, concrete or
ceramic tile. If floors are covered
with synthetic material, the relative
humidity should be at least 30%.

Electrical fasttransient/burst
EN 610004-4±2 kV for power
supply lines±1 kV for input/output lines

N/A

Mains power quality should be thatof a typical commercial or hospitalenvironment.

Surge
EN 61000-4-5± 1 kV differential mode
± 2kV common mode

N/A

Voltagedips,
shortinterruptions and voltage
variations on power supply
input lines
EN 61000-4-11< 5% UT(>95% dip in UT)
for 0.5 cycle
40% UT(60% dip in UT)
for 5 cycles
70% UT(30% dip in UT)
for 25 cycles< 5% UT(>95% dip in UT)
for 5 sec

N/A27

https://www.kadashika.jp

Electromagnetic
Guidance

envirnment

Mains power quality should be thatof a typical commercial or hospitalenvironment.
Mains power quality should be thatof a typical commercial or hospitalenvironment. If the user of HDR
requires continued operation during
power mains interruptions, it isrecommended that the HDR be
powered from an uninterruptible
power supply or a battery.

Power frequency (50/60
Hz) magnetic field
EN 61000-4-8

3A/m

3A/m

Conducted RF
EN 61000-4-6

3 Vrms150 kHz to 80 MHz

3V

Radiated RF
EN 61000-4-3

3 V/m
80 MHz to 2.5 GHz

3 V/mPower frequency magnetic fields
should be at levels characteristicof a typical location in a typicalcommercial
orhospital
environment.
PortableandmobileRFcommunication equipment should
be used no closer to any part of
HDR, including its cables, than
the recommended separation
distance calculated from theequation applicable to thefrequency of the transmitter.
Recommended
separation
distance:

to 800 MHz

for 80 MHz

for 800 MHz
to 2.5 GHz
Where P is the maximum output
rating of the transmitter in watts(W) according to the transmitter
manufacturer and d is therecommended
separationinmeters (m).
Field strengths from fixed RF
transmitters, as determined by anaelectromagnetic site survey,
should be less than thecompliance
level
in
each
b
frequency range.
Interference may occur in thevicinity of equipment marked
with the following symbol.

NOTE 1: At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people
a Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile
radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy.
To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should beconsidered. If the measured field strength in the location in which the HDR is used exceeds the applicable RF complianceabove, HDR should be observed to verify normal operation. If abnormal performance is observed, additional measures
may be necessary, such as reorienting or relocating the HDR.
b Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m28

https://www.kadashika.jp

Table 3. Recommended Separation Distance between Portable and Mobile RF Communications Equipment and HDR
PLEASE NOTE: HDR is intended for use in an electromagnetic environment in which radiated RF disturbances arecontrolled. The customer or user of HDR can help prevent electromagnetic interference by maintaining a minimum distance
between portable and mobile RF communications equipment (transmitters) and HDR as recommended below, according to
the maximum output power of the communications equipment.

Rated maximum output power of

Separation distance according to the frequency of the transmitter (m)the transmitter (W)150 kHz to 800 MHz

0.01

0.12

0.12

0.23

0.1

0.38

0.38

0.73

1

1.2

1.2

2.3

10

3.8

3.8

7.3

100

12

12

23

80Mhz to 800Mhz

800 MHz to 2.5 GHz

For transmitters rated at a maximum output power not listed above, the recommended separation distance d in meters (m) canbe estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of
the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1: At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection
from structures, objects, and people29

https://www.kadashika.jp