Page 34

5100 SERIES HIGH CAPACITY COMPRESSION
COLUMN FORCE TRANSDUCERSFEATURES• Compact Design• .15% Accuracy• Special Capacities to 2 Million lbf.
• 5pt Compression Calibration• Proprietary Modulus Compensated Strain Gages• Metric Capacities Available - Consult Factory

SPECIFICATIONS
Model

PARAMETERS

Capacity

U.S. lbf.

5100 SERIES HIGH CAPACITY COMPRESSION
COLUMN FORCE TRANSDUCERSDIMENSIONS
Model

5130

5135

5140

5160

5180

100K

200K

300, 400K

500K, 600K

750K, 1000K+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006

5130

5135

Capacity
Dimensions
1.

100K lbf
inch

200K lbf
inch

2.

R 1.99, 2.26

R 2.33

3.

2.00

3.00

4.
5.

2.38
.03

3.25
.03

6.

1.68

2.37

7.

3/8 -24.50 Deep

1/2 - 20.50 Deep

8.

R 12

R 12

5.00

7.00

5140
300K,
400K lbf
inch

9.00
R 2.78- 300K
R 3.00- 400K
3.50-300K
3.70-400K
3.95.03
2.94-300K
3.50-400K
1/2 - 20.75 Deep.
R 15-300K
R 16-400K

5160
500K, 600Klbf
inch

5180
750K,
1000K lbf
inch

R 3.29R 4.02

4.50

6.00

5.00
.03

6.50
.03

3.89

5.02

1 - 14.75 Deep

1 - 141.00 Deep

R 20

R 20

10.00

12.00

ACCURACY

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

TEMPERATURE
Range, Compensated, °FRange, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ ° F
Effect on Zero, % Rated Output/ ° C

OPTIONS
Additional bridges
Output adjustment to standard
Compression overload protection
Integral cable
Custom connectors

ELECTRICAL
Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, MegohmBayonet Connector

215350350
5000222151515350
350
350
350
350
350
5000
5000
5000
PT02E-10-6P (MS3112E-10-6P)

215350
350
5000

150
13
Steel150
96
SteelConnector Protectors

ACCESSORIESMating connector
Loading fixtures

MECHANICAL
Safe Overload Range, % Rated Capacity
Weight lb
Flexure Material

150
5
Steel

150
25
Steel

150
45, 46Steel

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Interconnect cable
Shunt calibration resistor
Instrumentation

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 35

Page 34

5100 SERIES HIGH CAPACITY COMPRESSION
COLUMN FORCE TRANSDUCERSFEATURES• Compact Design• .15% Accuracy• Special Capacities to 2 Million lbf.
• 5pt Compression Calibration• Proprietary Modulus Compensated Strain Gages• Metric Capacities Available - Consult Factory

SPECIFICATIONS
Model

PARAMETERS

Capacity

U.S. lbf.

5100 SERIES HIGH CAPACITY COMPRESSION
COLUMN FORCE TRANSDUCERSDIMENSIONS
Model

5130

5135

5140

5160

5180

100K

200K

300, 400K

500K, 600K

750K, 1000K+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+/-0.150+/-0.150+/-0.150+/-0.010+/-0.025+/-1.0+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006+15 to +115
-10 to +45-30 to +200
-34 to +90+/-0.003+/-0.006+/-0.003+/-0.006

5130

5135

Capacity
Dimensions
1.

100K lbf
inch

200K lbf
inch

2.

R 1.99, 2.26

R 2.33

3.

2.00

3.00

4.
5.

2.38.033.25.03

6.

1.68

2.37

7.

3/8 -24.50 Deep

1/2 - 20.50 Deep

8.

R 12

R 12

5.00

7.00

5140
300K,
400K lbf
inch

9.00
R 2.78- 300K
R 3.00- 400K
3.50-300K
3.70-400K
3.95.03
2.94-300K
3.50-400K
1/2 - 20.75 Deep.
R 15-300K
R 16-400K

5160
500K, 600Klbf
inch

5180
750K,
1000K lbf
inch

R 3.29R 4.02

4.50

6.00

5.00.036.50.03

3.89

5.02

1 - 14.75 Deep

1 - 141.00 Deep

R 20

R 20

10.00

12.00

ACCURACY

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

TEMPERATURERange, Compensated, °FRange, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ ° FEffect on Zero, % Rated Output/ ° C

OPTIONS
Additional bridges
Output adjustment to standard
Compression overload protection
Integral cable
Custom connectors

ELECTRICAL
Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm
Bayonet Connector

215350350
5000222151515350
350
350
350
350
350
5000
5000
5000
PT02E-10-6P (MS3112E-10-6P)

215350
350
5000

150
13
Steel

150
96
SteelConnector Protectors

ACCESSORIESMating connector
Loading fixtures

MECHANICAL
Safe Overload Range, % Rated Capacity
Weight lb
Flexure Material150
5
Steel150
25
Steel150
45, 46Steel

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Interconnect cable
Shunt calibration resistor
Instrumentation

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 35