Other products

Solar simulator for duration tests
� Simple test system with SOLAX
�■ Solar simulator for■duration tests

Simple test system withSOLAX*ModelSML-2K1A*ModelSXL-500V2*Lamp type(Qty)*Irradiance area

2kW Metal halide lamp
(1ea)
300×300mm*Lamp type(Qty)*Irradiance area

Solax
XC-500EFSS ( 1 ea)
φ100mm ／ 100×100mm*Irradiance

1kW/m ≦

*Irradiance

1kW/m ≦

*Non-uniformityof irradiance*Temporal instability*Adjustment rangeof irradiance*Weight≦±10%*Non-uniformityof irradiance*Temporal instability*Adjustment rangeof irradiance*Weight≦±5% / ≦±10%2≦±3%50～100%（Single step）
150kg

2

Solar Simulator≦ ±4%50～100%（Single step）

Solar Simulators for Cells/Modules

45kg■Consumables
Qty/system
*1

Model

Lamp type

50×50mm

XIL-01B50KP

140W Xenon lamp

80×80mm

XIL-05B80KP

500W Xenon lamp

1 ea

100×100mm

XIL-05B100KP

500W Xenon lamp

1 ea

156×156mm

XIL-05B156KP

500W Xenon lamp

1 ea

156×156mm

XIM-1B156KP

1000W Xenon lamp

1 ea

210×210mm

XIL-1B210KP

1000W Xenon lamp

1 ea

210×210mmXIM-3B210KP

3000W Xenon lamp

1 ea

300×300mm

XIM-3B300KP

3000W Xenon lamp

1 eaContinuous type solar simulator

SHR-50KA

2000W Metal halide lamp

20 ea

Solar simulator for duration test

SML-22KP

2000W Metal halide lamp

9 ea

Solar simulator for duration test

SML-2K1A

2000W Metal halide lamp

1 ea

Simple test system with SOLAXSXL-500V2500W Xenon lamp

1 ea

Solar simulators for Cells

1 ea

SERIC Ltd. is the only company in Japan to offer artificial solar
illumination lamp technology and can provide the optimum solar simulator
no matter what your requirements are.

Solar simulators for Modules

Others*1

Qty refers to the number of lamps required per replacement.

Sales┃
Customers include:

Universities/National laboratories/Automobile manufacturers/ Electronics

SERIC has supported thedevelopment of solar batteries
through artificial solar
illumination lamp technology
for over 25 years.

manufacturers/Chemical companies/Building material manufacturers
Usages include:

Photo voltaic cell tests/Duration tests/Evaluation of rocket parts

Contact
Please feel free to contact us with any inquiries.

E-Mail serinfo@seric.co.jp
TEL+81-3-3555-2261+81-3-6807-4811 ◎AM9：00～18：00 (Saturday, Sunday and National holiday is off)
+81-3-6807-4812
FAX+81-3-3555-2262 ◎Available 24h a day
Irifune, Tel,

Websitehttp://www.en.seric.co.jp/
https://www.en.seric.co.jp/

6-3-8-703, Irifune,Akasaka,Chuo-ku,
Minato-ku,
Tokyo TEL +81-3-3555-2261【Head office】3-1-2-505
Tokyo+81-3-3555-2262+81-3-6807-4811 FAX +81-3-6807-4812【Technical center】7-334-1 Shichiza-cho, Koshigaya, Saitama TEL +81-48-969-2155 FAX +81-48-969-2166
E-Mail serinfo＠seric.co.jp
http://www.seric.co.jp
https://www.en.seric.co.jp/

https://www.en.seric.co.jp/
http://www.en.seric.co.jp/

Our Three Passions┃ We take pride in contributing to the development of our customers through the groundbreaking technology born of our passion

1

2

3

Providing customer satisfaction

Guaranteeing the best after sales
service to our customers

Guaranteeing the lowest cost and themost compact devices

Spectrum distribution

SERIC SOLAR SIMULATORS for Cells

0301
Solar simulator

AM1.5

・Easy touch panel operation

Spectral coincidence
JIS C 893350×50mm

太阳模拟器

156×156mm

太阳模拟器□ 156 × 156 mm Solar simulator

Down shine
Flexible irradiance direction :Horizontal
Up shinecustomization according to module
02 Flexible
size, installation space etc

Solar simulator

自然太阳光
Actualsun light

太阳模拟器
Solar simulator

specification (grade) depending on
03 Flexible
your budget

AM1.5

Se
cifi
cation2 Si z e and we i ght may c hange wi thout pr i or noti c e

3 Ave r age l amp l i fe i s base d on n on-s t op ope r ati on

Spectral coincidenceJIS C 8912

SERIC offers a variety of solar
simulator grades from standard up toClass AAAp

 1 Al l sol ar si m ul ator s ha ve a fe e dbac k func ti on

SOLAR SIMULATORS for Modules

Spectral coincidence
JIS C 8912

□ 50 × 50 mm Solar simulator

SERIC

Irradiance

02

・JIS C 8912/8933 Class AAA・Angle of incident light : within 1°

Irradianceline up
01 ・Extensive
Irradiance area ：50×50mm～300×300mm

Spectrum distribution■Continuous type solar simulator
 ModelSML-50KA
 Irradiance area
W1.8×H1.5m
 Number of lamps2kW Metal halide lamp 20 ea
 Irradiance
1kW/m2
 Spectral characteristics AM1.5G
 Spectral coincidence
JIS C 8912 A
 Temporal instability
JIS C 8912 A
 Non-uniformity ofJIS C 8912 Airradiance Irradiation distance3m
 Direction of irradiance Horizontal

自然太阳光

Photo by SHINNIHON KAGAKU MFG.,LTD■ Solar simulator for duration test










Model
SML-22KP
Irradiance area
W1.8×H1.5m
Number of lamps
2kW Metal halide lamp 9 ea
Irradiance
1kW/m2
Spectral characteristics AM1.5G
Spectral coincidenceJIS C 8912 B
Temporal instability
JIS C 8912 A
Non-uniformity ofJIS C 8912 Cirradiance
Irradiation distance
1260mmDirection of irradiance
Down shine

太阳模拟器

Photo by SHINNIHON KAGAKU MFG.,LTD※ SERIC Ltd. offers both temperature and humidity test chamber s and light sources system .