Page 46

PREMIUM “S” TYPE(SSC) SERIES LOAD CELLSSCT SERIES PRECISION OVERLOAD
PROTECTED S-TYPE LOAD CELLSFEATURES• Premium Version of the Standard “S” Cell
• Designed for the Test Measurements Market• In-Line Loading• Sealed for Environmental Protection• Bayonet Connector for 500 lbf and above• 10’ Cable for 250 lbf and below• 3pt Tension Calibration• Proprietary Modulus Compensated Strain Gages

SPECIFICATIONS
Model

PARAMETERS

Capacity

SSC
50, 100, 250,U.S. lbf.
500, 1K
2K, 3K, 5K

ACCURACY

Static Error Band, % Rated OutputNonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output+/-0.030+/-0.030+/-0.030+/-0.010+/-0.030+/-1.0+/-0.050+/-0.050+/-0.050+/-0.010+/-0.030+/-1.0Range, Compensated, °FRange, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ °F
Effect on Zero, % Rated Output/ °C+15 to +115
-10 to +45-65 to +200-55 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-65 to +200
-55 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027

Rated Output, mV/V, Nominal
Excitation Voltage, VDC Recommended/Max
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm

33
10/15
10/15
350 +40/ -3.5
350 +40/ -3.5
350 +/- 3.5
350 +/- 3.5
5000 (@ 50 VDC)
PT02E-10-6P(MS3112E-10-6P)

TEMPERATUREELECTRICAL

Bayonet Connector

MECHANICAL

Safe Overload Range, % Rated Capacity
Weight lb
Flexure Material+/-150
0.5
Aluminum+/-150
1.2
Stainless Steel

Page 47

FEATURES• Overload Protected to 1000% of FS
in both Tension and Compression• High Accuracy-.02% of R.O.
• Integral Cable• 5pt Tension & Compression Calibration• Proprietary Modulus Compensated Strain Gages

DIMENSIONSCapacity
lbf.
50
100
250
500
1K
2K
3K
5K

1
2.00
2.00
2.00
2.00
2.00
2.00
2.00
2.50

2.48
.48
.48

.97
.97
.97
.97
1.47

SPECIFICATIONS
3.80.80.80

1.30
1.30
1.30
1.30
1.80

4
2.50
2.50
2.50
3.00
3.00
2.97
2.97
3.47

5
1/4 - 28 UNF -2B
1/4 - 28 UNF -2B
1/4 - 28 UNF -2B
1/2 - 20 UNF -2B
1/2 - 20 UNF -2B
1/2 - 20 UNF -2B
1/2 - 20 UNF -2B
5/8 - 18 UNF -2B

Model

PARAMETERS

Capacity

ACCURACY

U.S. lbf.
Metric N+/-0.030+/-0.040+/-0.030+/-0.010+/-0.025+/-1.0Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ °F
Effect on Zero, % Rated Output/ °C

0 to +125-15 to +50-10 to +175
-25 to +80+/-0.0010+/-0.0018+/-0.0015+/-0.0027

Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm

215350350
5000

ELECTRICALMECHANICALACCESSORIES

Output adjustment to standard
Integral cableSpecial connector
Metric Capacities
3pt Tension & CompressionCalibration

Mating connector
Loading fixtures
Interconnect cable
Shunt calibration resistor
Instrumentation

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

2.2 - 225
10 - 1000 N

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

TEMPERATURE

OPTIONS

SCTSafe Overload Range, % Rated Capacity
Weight lb
Flexure Material

DIMENSIONS
ModelSCT
2.2, 4.4, 11, 10, 20, 50,
Capacity 22, 55 lbf 100, 250 N 110, 225 lbf 500, 1000 N
inchmminch
mmDimensions
2.00
50.8
2.25
57.1
1.
.50
12.7
1.00
25.4
2.
.85
21.6
1.35
34.3
3.
2.50
63.5
3.0
76.2
4.
5.

1/4 - 28.37 Deep+/-1000
0.3
Aluminum

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

M6 X 1
9.4 Deep

1/2 - 20.54 Deep

M12 X 1.75
13.7 Deep