Torque Wrench Tester

Digital Torque Wrench Tester

DOTE4-G/-MD2
Ideal for using torque wrench calibration
Multicolor LED shows judgment status in three diﬀerent colors

1digiteight[kg]18205052pter

6k

DOTE100N4-G

Uniform loading device enhances the repeatability & accuracy
Models available to cover a torque range of 2 to 1000N・m
Units are selectable, cN・m/N・m/Kgf・cm/Kgf・m/lbf・in/lbf・ftm

Measurement data can be output by RS232C or USB
Connecting cable is sold separately.

.

17.EN.S

607-0624_EN.indd

2

2017/10/20

17:14:07Digital Torque Wrench Tester

Features

D

Horizontally fixed uniform loading device ensures consistent and highly accurate,
repeatable results

EC

Using the loading bar puts all the force on the eﬀective length point on the tool for thehighest accuracy.

*9

Newly designed handle is easy to operate even at maximum torquePole

Operator turning force remains low throughout the torque range.

Central display allows the operator to view direct read style torque wrenches easily
especially for larger size model
Operator position allows eﬀortless viewing of direct read tools.
Back PanelWheelPane

DOTE4-G-MD2: Motor Driven Loading Device

Suitable for bulk calibrations
Three step loading speeds to suit the operator
Includes an inching dial for small adjustments to direct readstyle torque wrenches
Motor drive units can be retroﬁtted to DOTE4-G tester

Complete w/MD Set
Model

DOTE20N4-G-MD2
DOTE50N4-G-MD2
DOTE100N4-G-MD2
DOTE200N4-G-MD2
DOTE500N4-G-MD2
DOTE1000N4-G-MD23HW

Retrofit Motor Driven Unit
Model

Applicable model

MD2-SET-S

DOTE20N4-G
DOTE50N4-G
DOTE100N4-G
DOTE200N4-G

MD2-SET-LDOTE500N4-G
DOTE1000N4-G

ExCo* Power cord will be provided in diﬀerent shapes depending on each country.
Contact to Tohnichi.

607-0624_EN.indd

U2017/10/20

17:14:22

Data Output Function
Enough test result data necessary for tool management can be output
Control Number - Torque Value - Unit - Hi/Lo setting pattern - Result - ID number* 99 memory mode that is used in previous model is also available.
Pole* default: serial No. of DOTE4-G

Tohnichi free software, ExRcv is available.
ExRcv Software

4-ＧThrough optional connecting cable,stored data can be output as Microsoft
Excel® format.
Plate

Optional Tohnichi Tool-Management software is available.
All measuring result can be uploaded and torque wrenches can becontrolled. Please contact Tohnichi for further details.

Pole Holder Plate

Certificate under ILAC MRA

Calibration certiﬁcate valid by Mutual
Recognition Arrangements of ILAC andAPLAC is optionally available.
The ILAC Mutual Recognition Arrangement is recognized as meeting international
standards of competence, and the certiﬁcate under ILAC-MRA provides additional
conﬁdence since it certify the torque to be traced directly to National standard.

Common Specifications
Wheel Handle

PanelDisplay

Black mask LCD, White-Red-Blue

Measurement mode

PEAK/RUN

Data memory

1000 data

Hi/Lo setting pattern

10 setting pattern, A to J

Statistical processing

Number of samples/max./min./mean value

Zero adjustment

AutomaticReset

Manual 0.0/Auto 0.1-5.0 sec.

Data output

RS232C compliant, USB B type for serial communication

Operating condition

0-40℃ (No condensation)

Power supply

AC100-240V 50/60Hz

Up to 10 Hi/Lo settings can be programmed for easy retrieval
readHi/Lo settings can also be programmed remotely through a PC.
With separately prepared barcode reader, Hi/Lo settings can be done by command input.
Application Example of Hi/Lo Setting by Barcode Reader
During calibration

ExampleCommand inputry.

Hi, Lo value setting commandHi607-0624_EN.indd

4Lo

OK judgment

NG judgment

2017/10/20

17:14:22

Speciﬁcations
DOTE20N4-G(-MD2) - 200N4-G(-MD2)

DOTE500N4-G(-MD2), 1000N4-G(-MD2)
Motor Drive(DOTE4-G-MD2)

L
Max. Eﬀective Length L’L

Motor Drive

Max. Eﬀective Length L’

Panel- Power- Power Jack- D-Sub 9 Pin (male)- USB (B type)

Pole Holder Plate

Serial Number

Plate

Plate

112

Panel- Power- Power Jack- D-Sub 9 Pin (male)- USB (B type)

150

Memory/Reset
PoleMemory/ResetPole

L1Wheel Handle

H1H2
2215

H1

H2

H

H

Wheel Handle

30°

B

B1
346

252

294

B

Serial Number

192

30°

Inlet Drive

710

(153)

Inlet Drive

B2(230)(239)(DOTE4-G-MD2)

Φ32

Φ50

Accuracy +/-1%+1digit
cN・m

Model

N・mkgf・cmkgf・mlbf・inlbf・ft

Min.- Max.

1 digitMin.- Max.

1 digit Min.- Max.

1 digit Min.- Max. 1 digit

Min.- Max.

1 digit Min.- Max. 1 digit

DOTE20N4-G

200-2000

0.2

2-20

0.002

20-2000.02--

18-1800.02--

DOTE50N4-G-

-

5-50

0.005

50-5000.05--

44-4400.05

3.6-360.005

DOTE100N4-G--

10-1000.01 100-1000 0.1--

88-8800.1

7.3-73

0.01

DOTE200N4-G--

20-2000.02 200-2000 0.2--

170-1700

0.2

15-1500.02

DOTE500N4-G--

50-500

0.05

-

-

5-50

0.005 440-4400

0.5

36-3600.05

DOTE1000N4-G--

100-1000

0.1

-

-

10-1000.01 880-8800

173-730

0.1Dimensions [mm]
Model

L

H

B

L1

B1

B2------------

810

518

90

DOTE20N4-G
DOTE50N4-G

506

DOTE100N4-G
DOTE200N4-G

339

756

391

1276

710

DOTE1000N4-G 1856

775

DOTE500N4-G

267
269
394

1200

666

Optional Accessories
PrinterHex Adapter
Part No.
285
286
287
288

Size

3/8-7・8・9
1/2-16・18・21
1/2-17・22・27
1/2-19・24・30

Model

EPP16M3
1408
382

150

H1113

162

Standard Accessories
Adapter (inlet-outlet)

H2#296 DA3-2 (1/4-3/8)

171

296
297
298
299

Model

DA3-2DA4-3DA6-4DA8-6660

2019.51120

5025.41750

52
AC Adapter

10・13・1912・14・17

174#297 DA4-3 (3/8-1/2)

17・22・27 / 19・24・30

238#298 DA6-4 (1/2-3/4)

22・27・29 / 30・32・36

247#299 DA8-6 (3/4-1”)

34・41 / 46・50

Description

EPP16M3BA-6

Battery Pack

Connecting Cable
Description
Model
383 DOTE4 - PC (D-Sub 9pin)
385 DOTE4 - PC (USB B type)Model
BP-100-4385

Calibration Kit

14 N・m
70 N・m220 N・m
750 N・mProduct movies are available. Visit Tohnichi TV.

https://www.youtube.com/c/TohnichiTV[kg]18#297 DA4-3 (3/8-1/2), #277 (1/4-1/2)

Inlet-Outlet Sq. Max. Torque

1/4 - 3/8
3/8 - 1/2
1/2 - 3/4
3/4 - 1”

12.7

Weight410

169383

Part No.

9.5

Hex Adapter [mm]#296 DA3-2 (1/4-3/8)

Thermal line dot printerRoll paper
Printer cable

Down Adapter

Max. available
eﬀective length
L’ [mm]

Inletdrive

Model
DOTCL-S1
DOTCL-S2
DOTCL-S3
DOTCL-L3
DOTCL-L4

Applicable Models
DOTE20N4, 50N4-G
DOTE100N4-G
DOTE200N4-G
DOTE500N4-G
DOTE10000N4-G

TOHNICHI MFG. CO., LTD.
URL
http://www.global-tohnichi.com
E-mail overseas@tohnichi.co.jp
N.V. TOHNICHI EUROPE S.A.
E-mail tohnichi-europe@online.be
TOHNICHI AMERICA CORP.
URL
http://www.tohnichi.com
E-mail inquiry@tohnichi.com
TOHNICHI SHANGHAI MFG. CO., LTD.
URL
http://www.tohnichi-sh.com
E-mail sales@tohnichi-sh.com

The appearance and speciﬁcations are subject to change for improvement without prior notice.
All right reserved. No reproduction or republication without written permission.
TOHNICHI Mfg. CO., LTD. All Rights Reserved.

607-0624_EN.indd

17000.10.17.EN.S

2017/10/20

17:14:06