Page 52

9150 HIGH PRECISION
INDICATOR

9150-X-300 HIGH PRECISION
INDICATORFEATURES• 24-bit Resolution• 10-point Linearization• Linearity of 0.005% of Full Scale• Sensitivity of +/-4.5 mV/V
• 2 Channel Transducer Input (Optional)• Software Selectable Filters• RS-232, RS-485 or USB Computer Interface• Stores Calibration Data for up to 25 Load cells

FEATURES• Ratiometric measurement measures true mV/V
• Resolution 0.00001 mV/V
• Display mV/V or physical units• Stores calibration data for up to 25 load cells• 10-point linearization• Special high performance unit
• Linearity of 0.003% of full scale• Sensitivity of +/-4.5 mV/V
• 1 or 2 channel transducer input and display• RS-232, RS-485 or USB Computer Interface• Software selectable filters SPECIFICATIONSSPECIFICATIONS

9150-1 Single Channel
9150-2 Dual Channel

Model

PARAMETERSPARAMETERS

Power Requirements
Excitation
Current Drive
Push Button Shunt
Internal Shunt Resistor
Display Values
Calibration Method
Push Button Tare
Sensitivity Adjust
Accuracy
CMRR

115 VAC, 60Hz/ 230 VAC, 50Hz
5VDC or 10 VDC software/auto selectable
180 mA at 5 VDC or 10 VDCYes
1 standard, 2 optional. Rear panel selectable-999,999 through 999,999
Shunt, mV/V, Known Load
100% of range--display and analog output
1 mV/V to 4.5 mV/V
0.01% of full scale +/- digital count
115 dB

Number of Channels

9150-1 Single, 9150-2 Dual

Full Scale Output
Linearity
Scaling
Frequency Response
Output Source+/- 10 VDC nominal +/- 0.5 VDC
0.02% of full scale
Gain and offset selected software
Approx. 15 Hz (plus filtering)
Any displayed quantity

Full Scale Output
Linearity
Scaling
Frequency Response+/- 8.14 VDC +/- 0.25 VDC
0.02% of full scale
Fixed with no tare, transducer channel onlyApprox. 1.5 kHz

Inputs--Interface
Outputs--Quad Limits

4 optically isolated inputs
4 isolated solid state switches

RS-232, RS-485
USB

Standard, Optional
Standard

Auto I.D. and Setup

Standard

TRANSDUCER CHANNELS
D/A ANALOG OUTPUT

DIRECT ANALOG OUTPUTDIGITAL I/O

SERIAL INTERFACEAUTO TRANSDUCER I.D.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

The Model 9150-X-300 is a high performanceversion of the standard Model 9150 featuring
higher accuracy, stability and lower temperatureerrors.

9150-1-300 Single Channel
9150-2-300 Dual Channel

Model

PERFORMANCEPERFORMANCEPower Requirements
Excitation
Current Drive
Push Button Shunt
Internal Shunt Resistor
Display Values
Calibration Method
Push Button Tare
Sensitivity Adjust
Accuracy
CMRR

115 VAC, 60Hz/ 230 VAC, 50Hz
5VDC or 10 VDC software/auto selectable
180 mA at 5 VDC or 10 VDCYes
1 standard, 2 optional. Rear panel selectable-999,999 through 999,999
Shunt, mV/V, Known Load
100% of range--display and analog output
1 mV/V to 4.5 mV/V
0.003% of full scale +/- digital count
115 dB

Effect on Zero
Effect on Output

0.005%/C
0.005%/C

Number of Channels

9150-1-300 Single, 9150-2-300 Dual

Full Scale Output
Linearity
Scaling
Frequency Response
Output Source+/- 10 VDC nominal +/- 0.5 VDC
0.02% of full scale
Gain and offset selected software
Approx. 15 Hz (plus filtering)
Any displayed quantityFull Scale Output
Linearity
Scaling
Frequency Response+/- 8.14 VDC +/- 0.25 VDC
0.02% of full scale
Fixed with no tare, transducer channel onlyApprox. 1.5 kHz

Inputs--Interface
Outputs--Quad Limits

4 optically isolated inputs
4 isolated solid state switches

RS-232, RS-485
USB

Standard, Optional
Standard

Auto I.D. and Setup

Standard

TEMPERATURE

TRANSDUCER CHANNELS
D/A ANALOG OUTPUT

DIRECT ANALOG OUTPUTDIGITAL I/O

SERIAL INTERFACEAUTO TRANSDUCER I.D.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 53