NICHIRYO PRODUCT LINEUPPIPETTESpecifications

Autoclavable Micro Pipette

Cat. No
00-NLT-200-NLT-1000-NLT-2000-NLT-10000-NLT-200

Innovation in “Fatigue-Free” & “Comfort”

Volume Setting

Volume range(µL)
0.1 – 2
0.5 – 102 – 20
10 – 100
20 – 200

Digital Setting

00-NLT-1000

Increments(µL)
0.002
0.01
0.02
0.1
0.2

100 – 1000

1

Usable tips
BMT2-UT/UTWR/UTWRB、FUT/FUTRB
BMT2-SSW/SSWR/SSWRB 、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB*See p.21-22 for tip selection.

Accuracy and Precision
Cat. No00-NLT-2NewFeature

00-NLT-1000-NLT-20
Keeps the same durabilit y as
Nichipet Premium and reduces
less 70% stroke loading atmaximum. Mitigates fatigue andstress from heavy workload.

This will not be worn out within
the clean bench work and beenlarged easy to read thevolume.

Accuracy（%）±12.0*±5.0±3.0±4.0*±1.0±1.0±5.0±1.0±1.0

Volume(µL)
0.212151021020Precision
（%）
≦6.0*
≦2.5
≦1.0
≦3.0*
≦1.0
≦0.5
≦3.0
≦1.0
≦0.4

Cat. No00-NLT-10000-NLT-20000-NLT-1000

Volume(µL)
1050100
20100200100500
1000

Accuracy（%）±2.0±1.0±0.8±1.0±0.8±0.8±1.0±0.8±0.7

Precision
（%）
≦1.0
≦0.3
≦0.3
≦0.5
≦0.3
≦0.2
≦0.5
≦0.3
≦0.2

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NLT-2 below 0.2μL and NLT-10 below 1μL depend on the sampling skill and the sampling condition where pipettes are used.

Nichiryo's quality starter packages

S-PAC

New material used for 4 notcheslock improves its strength which
minimizes the lever shifting andavoids drifting and wearing.

ceramic plunger
Non corrosive ceramic used inthe plunger.
( 100µL – 1000µL )■ Package type of 3 selected Pipettes with racked -Tips.
■ Ready-to-start packages with 3 sizes ofpipettes to perform highAC and CV pipetting.
■ 3 packages are available to cover wide range of volume.

Material used has hightolerance to solvents and to drop
impact. It is fully
autoclavable as well.
Nozzle top has highly abrasion
tolerant. ( 2µL – 1000µL )Hyper blower system improvesdispensing efficiency, especiallyon 2µL and on 10µL dispensing.

Nichiryo Tube Opener (code:00-NTO)

S-PAC

3 pcs

Racked Tip

Tube Opener

3 or 4 racks

3 pcs“Easier” and “safer” to open tube lid with holding pipette■ Attached on grip end, 0.5, 1.5 mL tube

and screw cap can be opened easily.
＊C o m p a t i b l e w i t h N i c h i p e t P r e m i u m LT( 2-1000µL ) and Nichipet EXⅡ ( 2-1000µL )● What's new!
■The durability still is same as Nichipet Premium has. Stroke loading of Nichipet Premium LT

Shortened to releasetension to your thumb.

becomes less 70% at maximum than before.
■ Hand temperature doesnʼt permeate through the body of the pipettewhich works to avoid affecting volume measurement. ( Patented )＊All features of Nichipet Premium are available on Nichipet Premium LT.
Since the volume indicator is made of glass, ultraviolet germicidal of the clean bench will not make it cloud.
From prolonged irradiation, the surface layer of plastic may deteriorate or discolor, but its performance is not affected.

09

page.
www.nichiryo.co.jp/en/

Shortened
the distance tooperate smoothly.

Nichipet Premium LT S-PACの構成
Description

Cat. No

Nichipet Premium
A
00-SP-AL-PACの構成
Nichipet
EXLTSS-PAC

Content
NLT-10、
NLT-100、
NLT-1000
BMT2-SSWR（0.5〜10μL）、
BMT2-SGR（2〜200μL）、
BMT2-LGR（100〜1000μL）、
NTO

Nichipet Premium LT S-PAC B

00-SP-BL

NLT-20、
NLT-200、
NLT-1000
BMT2-SGR（2〜200μL）
×2、
BMT2-LGR（100〜1000μL）、
NTO

Nichipet Premium LT S-PAC H

00-SP-HL

NLT-2、NLT-10、
NLT-100
BMT2-UTWR（0.1〜10μL）
×2、
BMT2-SSWR（0.5〜10μL）、
BMT2-SGR（2〜200μL）、
NTO＊visit our web-site for part lists.

page.
www.nichiryo.co.jp/en/10