NICHIRYO PRODUCT LINEUPDISPENSER

Variable Volume Bottle Dispenser

Autoclavable Bottle-top Dispenser

ACCUPENSER JR.

DISPET EX■The Post with scale can be laid down without altering volume settings.

■ Autoclavable（ 121℃ for 20 minutes ）■The precision parts of the glass are constructed and protected on the inside.

■ 3 sized adapters for fitting with various types of bottles.

■The liquid contact parts are chemically resistant.

■ Chemical resistant materials are used for liquid contact parts.

■The parts can easily be dismantled and changed.

■ Adjustable Inlet tube for various types of bottles.

■ Liquid can be added with the funnel without changing the setting of■ Liquid returning mechanism for minimizing the waste of reagents.

the device.

■The wetted parts are chemically resistant.

■The extension tube with tip allows dispensing to be done without moving

Borosilicate glass,fluorocarbon polymers,Ceramic,Platinum-Iridium

the device or sample bottle. Please use a PTFE ( Fluorine resin ) tube set
with tip ( sold separately ) for strong acids and organic solvents.
■ Autoclavable without disassembling■The wetted parts are chemically resistant.

Borosilicate glass,Glass,fluorocarbon polymers＊Tube:Glass,SiliconSpecifications
Specifications
Cat. No00-AJ-100-AJ-200-AJ-500-AJ-10

Volume range(mL)
0.2 – 1
0.4 – 2
1–52 – 10

Increment(mL)
0.05
0.1
0.1
0.25

Accuracy(%)±1.0±1.0±1.0±1.0

Precision(%)
≦0.1
≦0.1
≦0.1
≦0.1

Size of provided bottle
250mL
250mL
500mL
1L

Width(mm)×high(mm)75 × 210（165）75 × 210（165）
100 × 210（165）
146 × 210（165）＊The bottle has treads compatible with 500mL and 1L bottles for free combinations. ( ) indicates the height when the scale post is lying down.

Cat. No00-DPX2-2000-DPX2-5000-DPX2-100
00-DPX2-25000-DPX2-50000-DPX2-1000

Increment（mL）
0.05
0.1
0.20.5
12Accuracy
（%）
±0.5
±0.5
±0.5±0.5±0.5±0.5

Precision
（%）
≦0.1
≦0.1
≦0.1
≦0.1
≦0.1
≦0.1

ThreadAttached adapters

GL-45

GL-28、GL-32、GL-38

AdapterCat. No00-DPX-003A00-DPX-003B

Variable Volume Bottle-top Dispenser

Volume range（mL）
0.2 – 2.00.5 – 5.0
1.0 – 10.0
2.5 – 25.0
5.0 – 50.0
10.0 – 100.0

Size
32 / 28mm
45 / 32mm

Cat. No00-DPX-003C
00-DPX-003DSize
45 / 38mm
45 / 40mm＊Numbers right side of ʻ/ʼ are caliber size of the bottles.

DISPET■ Can be attached directly to all standard reagent bottles.

Bottle-top Dispenser■ Comes with an adapter ( ø22mm ) for direct attachment to smaller reagent bottles.

DISPET Mini

( 100mL> ) ( not DP-5 and DP-10 )■ Light weight and compact design for easy handling.
■ Liquid contact parts are chemically resistant material.
■ Extra tube ( 220mm ) is included and can be cut

into a more convenient size.
■The wetted parts are chemically resistant.

■ Suitable for repetitively micro volume dispensing.

Borosilicate glass,fluorocarbon polymers,Ceramic,Platinum-Iridium■ Light and compact model for easy-to-carry-out.
■ Chemical resistant materials are used for liquid contact parts.
■The wetted parts are chemically resistant.

Specifications
Cat. No00-DP-1B
00-DP-2B
00-DP-5B
00-DP-10B

Borosilicate glass,fluorocarbon polymers,Platinum-Iridium
Volume range(mL)
0.2 – 1
0.4 – 2
1–5
2 – 10Increment(mL)
0.05
0.1
0.1
0.25

Accuracy(%)±1.0±1.0±1.0±1.0

Precision(%)
≦0.2
≦0.2
≦0.2
≦0.2

Amber bottle(option)
200mL
200mL
350mL
1L

Width(mm)×high(mm)59 × 255
59 × 255
66 × 290
98 × 350＊Bottles are available as optional parts.

＊The threads for all bottle sizes are the same.
＊Please add "B" at the end of Cat.No when a bottle is needed.

AdapterCat. No
DP-1118
DP-1119
DP-112227

page.
www.nichiryo.co.jp/en/

Specifications
Size
18mm
19mm
22mm

Cat. No
DP-1124
DP-1125Size
24mm
25mmDP-1128

28mm

Cat. No
DP-1133
DP-1138
DP-1145

Size
33mm
38mm
45mm

Cat. No00-DPM-250
00-DPM-1000
00-DPM-2000

Volume range（µL）
100 / 250
500 / 1000
1000 / 2000

Accuracy（%）±2.0±1.0±1.0

Precision
（%）
≦0.4
≦0.2
≦0.2

＊visit our web-site for part lists.

Thread
GL-28
GL-28
GL-28

page.
www.nichiryo.co.jp/en/28