Genius 3051

Nitrogen & Air Generator for Shimadzu LC-MS
Part Number : 10-6360

Service Kit : See Reverse

Your local gas generation partner

Description
Engineered specifically for the Shimadzu LCMS-8045, LCMS8050, LCMS-8060 and LCMS-9030 instruments, the Genius 3051
provides one output of phthalate free, high purity nitrogen at up
to 27 L/min and one of dry air at up to 29 L/min. This will provide
enough high purity gas to support these instruments even whenbeing operated in 'high flow gas delivery mode' and for some uses
up to two instruments in 'standard gas delivery mode.
The Genius 30 series models have the highest capacity ofcompressor based systems Peak manufacture and feature a totalof four independent high-performance compressors enabling
supply of higher flow analytical instrument grade nitrogen.

Applications)

Shimadzu LC-MS 9030)

Up to 2 x Shimadzu LCMS-8045 in standard gas delivery mode)

Up to 2 x Shimadzu LCMS-8050 in standard gas delivery mode)

Up to 2 x Shimadzu LCMS-8060 in standard gas delivery mode)

Shimadzu LCMS-8045 in high flow gas delivery mode)

Shimadzu LCMS-8050 in high flow gas delivery mode)

Shimadzu LCMS-8060 in high flow gas delivery mode)

Shimadzu tested and approved solution for LCMS-9030)

Shimadzu LCMS-8060NX in high flow gas delivery mode)

Specifically designed for LCMS-8045, LCMS-8050 and LCMS8060)

Self-contained solution with integrated compressors, no needfor an external air supply)Supports LCMS "High flow gas delivery" mode)

Highly economical source of nitrogen/dry air with low lifetime
running costs)

Latest generation high performance compressors for longer
service interval)

Easy to use, gas at the push of a button)

Wheels make it easy to manoeuvre generator around the lab)

12 month comprehensive on-site warranty

Key Features

850.00 mm

600.00 mm

1322.00 mm

1322.00 mm
850.00 mm

600.00 mm

Genius 3051 Dimensions

Copyright 2020 Peak Scientific - Rev - Genius 3051 Rev 8 RSID 1458 EN - 30/06/20

Technical Specifications

Genius 3051

Max Flow Rate

27 L/min (0.91 cfm) N2 / 29 L/min (0.70 cfm) Dry Air

Max Pressure

100 psi / 690 kPa

Max Relative Humidity

80% Non-Condensing

Max Altitude2000 Metres

Air Dewpoint-20°C (-4°F)

Nitrogen Purity≤2% OxygenParticles

None

Gas Outlets

2x 1/4” BSPP

Phthalates

None

Suspended Liquids

None

Operating Temperature

5ºC - 35ºC / 41ºF - 95ºF

Electrical Requirements

230v single phase supply, 50/60Hz 12A

Power Consumption

2760 watts

Generator Dimensions (HxWxD)
Generator Weight1322 x 600 x 850 mm/ 52.0 x 23.6 x 33.4 in
216 Kg (476lbs)

Ordering Information
Part Number

10-6360

Annual Service
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan
Step-Up Transformer For Use With
208V Electrical Supply06-3210

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2020 Peak Scientific - Rev - Genius 3051 Rev 8 RSID 1458 EN - 30/06/20