ORBIT 60 SERIES Condition Monitoring
ModuleDatasheet

Bently Nevada Machinery Condition Monitoring

145M9028 Rev. A

Description
The Condition Monitoring Module (CMM) listens to all
information within the system, including all measurements,waveforms, digital transducer signals, system controls, status
information, system configuration information, process datafrom external systems, and alarm and events logs. It onlylistens, with no capability to write, allowing interface toSystem 1 over the business networks, with no risk to theprotection system.
Each module occupies two slots within the system. Placing
multiple CMM modules allows the connection of two
independent System 1 clients to the Orbit 60 System. Data istransferred to System 1 continuously, but in the event theconnection is lost, non-volatile storage buffers historical data
until the information is off-loaded to the host software.
System 1 can configure the CMM module to extract additional
measurements and waveforms from system sensor data.
Without System 1, the customer can use the CMM module tocollect data to diagnose machinery issues when an alarm
event occurs in the hardware.
Each module has two independently configurable RJ-45
Ethernet ports. OK LEDs indicate proper operation and LINK
LEDs show good communication to the system.

ORBIT 60 SERIES Condition Monitoring Module
Datasheet

145M9028 Rev. A

Condition Monitoring Module

Environmental Limits
ChassisOperatingTemperature
Range

Condition Monitoring Module (CMM)
Power Consumption
Maximum

14.2 W

Typical10.5 W(indoor useonly)

Data Communications
2 Ethernet
ports - utilityor rear side

Independent Ethernet ports
1000/100/10 Base-T Autonegotiation

Connector

RJ-45

Cable Length

100 meters (328 feet) max

3U Chassis:-30°C to +70°C(-22°F to 158°F)
6U Chassis:-30°C to +65°C(-22°F to 149°F)

ModuleTemperature
Rating Certification-30°C to +70°C(-22°F to 158°F)
You must still meet
the Chassis
OperatingTemperature Range
defined above.

LEDs
Module OK LED

Indicates when the module isfunctioning properly

Storage
Temperature
Range-40°C to +85°C(-40° F to 185° F)

LINK LED

indicates when the module iscommunicating to the rest ofthe system

Relative
Humidity

0% to 95% rH non-condensing
operating and storage

Vibration

Without Isolators:
0 g to 0.35 g @ 57-500 Hz

PhysicalRequired RackSpace

2 Slots

With Isolators:0 g to 5 g @ 57-500 Hz.
Shock2” Incline Drop

Altitude< 2000 m (6,562 ft)
Higher altitudes are
possible but are site
specificapplications.
Contact Bently
Nevadasupport if you
require higheraltitudes.

2/6

Pollution
DegreePollution Degree 2Installation
Category

Category II

ORBIT 60 SERIES Condition Monitoring Module
Datasheet

145M9028 Rev. A

Verify that temperature ratings on thewiring cables match the operating
temperature range.

CAUTIONLOCATION TEMPERATURE
AND HUMIDITY
If you install the hardware in alocation where temperatures may
exceed 40° C (104° F) or inexcessive humidity, you should
consider supplying environmentalcontrols. High temperatures will
reduce the operational life of thesystem.

3/6

ORBIT 60 SERIES Condition Monitoring Module
Datasheet

145M9028 Rev. A*MaritimeCompliance andCertifications

ABS Rules for Condition of Classification,
Part 1

FCC

ll

This device complies with part 15 of theFCC Rules. Operation is subject to thefollowing two conditions:ll

Steel Vessels Rules
Offshore Units and Structures* Approvals pending

Functional Safety

This device may not cause harmfulinterference.
This device must accept any
interference received, includinginterference that may cause
undesired operation.

This component is non-interfering withthe safety system. The system SIL 2certification does not require thiscomponent be SIL certified.

Hazardous Area Approvals

EMC

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

European Community Directive:
EMC Directive 2014/30/EU
Standards:

For additional technicaldocumentation, please log in to
bntechsupport.com and access theBently Nevada Media Library.

EN 61000-6-2; Immunity for
Industrial Environments
EN 61000-6-4; Emissions forIndustrial Environments

Electrical Safety

cNRTLusEuropean Community Directive:

Class I, Zone 2: AEx/Ex ec nC IIC T4 Gc;
Class I, Zone 2: AEx/Ex nA nC IIC T4 Gc;
Class I, Division 2, Groups A, B, C, D T4;
Class I, Division 2, Groups A, B, C, D T4 (N.I.);

LV Directive 2014/35/EU
Standards:
EN 61010-1;
EN 61010-2-201;

T4 @ Ta= -30˚C to +70˚C (-22˚F to +158˚F)

ATEX/IECEx

RoHS

II 3 G
Ex ec nC IIC T4 Gc
Ex nA nC IIC T4 Gc

European Community Directive:
RoHS Directive 2011/65/EU

T4 @ Ta= -30˚C to +70˚C (-22˚F to +158˚F)

Cyber Security
Designed to meet IEC 62443-4-2

4/6

ORBIT 60 SERIES Condition Monitoring ModuleDatasheet

145M9028 Rev. A

Ordering Information
For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).
For additional technicaldocumentation, please log in tobntechsupport.com and access theBently Nevada Media Library.

Condition Monitoring Module
Ordering Option

Description

60R/CMM01-AAA-B
AAA – Hazardous Area Certifications
00

No Hazardous Area01

CSA/NRTL/C (Class I, Div 2)

02

Multi (CSA, ATEX, IECEx)

XXXCountry Specific Approvals

B – SIL Level
0

No SIL

5/6

ORBIT 60 SERIES Condition Monitoring ModuleDatasheet

145M9028 Rev. A

Copyright 2023 Baker Hughes Company. All rights reserved.

Bently Nevada and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes business, in the UnitedStates and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company. All other product andcompany names are trademarks of their respective holders. Use of the trademarks does not imply any affiliationwith or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

6/6