Simplifying Progress
PFAS — Analysis Arium® Sterile Plus Filter
SampleDetection threshold

Detected Concentration

UnitMethodPFBA50

Under detection thresholdng/L (ppt)

MS-00473871; APFPeA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFHxA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFHpA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFNA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; APFDA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFUnDA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFDoDA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFTrDA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFTeDA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFHxDA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFBS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFPeS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFHxS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFHpS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFNS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFDS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFUnDS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFDoDS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFTrDS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

4:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

6:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

8:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

10:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

N-MeFOSAA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

N-EtFOSAA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; PV

8:2diPAP

0.5

Under detection thresholdng/L (ppt)

MS-00473871; PVPFECHS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; PV

Execution and Analysis Procedure
The water analysis was executed by TÜV Rheinland Energy & Enviroment GmbH, an internationally recognized testing laboratory for special analytics,
based on following measurement method: MS-0047387 Rev. 0, in accordance with DIN 38407-42, 2011-03. Relative expanded measurement
uncertainty (k=2): 50 %.
A= The method has been accredited.
PV= The method has been partially validated.
The tests were performed with the Arium® Sterile Plus Filter attached to an Arium® Mini Plus, fed with tap water. The Sterile Plus Filter was flushed with
5 liters of ultrapure water before the sample was collected.

SampleDetection threshold

Detected Concentration

UnitMethod

PFOA (linear)

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFOA (branched)

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFOA (total)

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFOS (linear)

0.5

Under detection thresholdng/L (ppt)

MS-00473871; APFOS (branched)

0.5

Under detection thresholdng/L (ppt)

MS-00473871; APFOS (total)

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

Execution and Analysis Procedure
The water analysis was executed by TÜV Rheinland Energy & Enviroment GmbH, an internationally recognized testing laboratory for special analytics,
based on following measurement method: MS-0047387 Rev. 0, in accordance with DIN 38407-42, 2011-03. Relative expanded measurement
uncertainty (k=2): 50 %.
A= The method has been accredited.
PV= The method has been partially validated.
The tests were performed with the Arium® Sterile Plus Filter attached to an Arium® Mini Plus, fed with tap water. The Sterile Plus Filter was flushed with
5 liters of ultrapure water before the sample was collected.

Germany
Sartorius Lab Instruments
GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0
For further information, visit	
sartorius.com

Specifications subject to change without notice.
© Sartorius Lab Instruments GmbH & Co. KG.
Status: 06 | 2024

USASartorius Corporation565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906