Your local gas generation partner

Product catalogueA

gas generation brandDiscover more at www.peakscientific.com

Our Promise
Our commitment is to provide yourlaboratory with a reliable gas generation
solution that delivers long term value. We
achieve this through exceptional productdesign, and direct global on-site support
that is unmatched in our industry.
We take a vested interest in youroutcomes and strive to be a valued partner
in your success by delivering exceptional
service over the life of the product.

Contents
1.0 Our story

4

2.0 Horizen

8

3.0 Genius

12

4.0 Solaris18

5.0 Infinity

22

6.0 Precision26

7.0 Specialist Solutions

36

8.0 i-FlowLab

38

9.0 PEAK Protected421.0 Our Story

About UsPEAK Scientific is a leading innovator in the design, manufacture and support of high performance gas generators
for analytical laboratories. Established in the UK in 1997 near Glasgow (Scotland), where our corporate Headquarters
and high-tech manufacturing and R&D facilities reside, PEAK Scientific boasts a significant local presence on every
continent – including major operations in North America, Europe, China and India.
With over two decades of experience in pioneering reliable gas generator technology, PEAK Scientific develops
market-leading nitrogen, hydrogen and zero air systems mainly for the fields of LC-MS (Liquid Chromatography-Mass
Spectrometry) and GC (Gas Chromatography).
What differentiates us is our world-class technical support and on-going service care throughout the generator’slifespan, wherever you may be in the world.

4

The PEAK Way
PEAK Scientific is a family owned business and, as such, people are at the core of our unique product offering. Eachgenerator is designed, assembled, tested, maintained and serviced by highly trained and dedicated professionals. The
importance placed on people at PEAK Scientific is consistent throughout our organization. It is the reason why we arecommitted to ongoing staff training and a policy of continual improvement in our engineering, manufacturing and field
service capabilities - delivered by over 650 employees worldwide.
Our values are structured around our colleagues, our customers and our service, and while our approach andatmosphere within the business is fun, friendly and informal, our priority is always on delivering successful outcomes forour customers.

Manufacturing Excellence
Our products are the result of our meticulous Research and Development culture developed over 25 years of being at
the forefront of the gas generator market. We pride ourselves on the utmost care taken to assess specific applicationneeds prior to designing and rigorously testing new products.
As we have expanded, so have our R&D capabilities, both technically and in terms of know-how, to the point where
we are better equipped than ever to meet changing market demands. Implementing the latest manufacturing
technologies and philosophies ensures that PEAK continues to set new standards for product quality, responsivenessand efficiency – all resulting in a better value for you.
Operating from our ISO 9001 accredited manufacturing center of excellence in the UK, every PEAK Scientific generator
is designed and tested to ensure compliance with the most internationally recognized safety and Electromagnetic
Compatibility Standards.

51.0 Our Story

Why a gas generator?
PEAK Scientific offers you a practical and cost-effective alternative to pressurized gas cylinders,
dewars or bulk storage of laboratory gas. Traditional sources of gas incur ongoing delivery,
administrative and rental costs, all of which impact on business revenue or facility budgets.
A PEAK generator provides you with a dependable, easy to use, on-demand gas solution without
the safety concerns and hassle that come with managing, stocking and restocking supplies whenusing gas cylinders or other methods of bulk gas supply. What’s more, while the price of delivered
gas is subject to volatility as well as delivery delays, a generator from PEAK Scientific represents astable and dependable long-term investment.

6

Convenient

Consistent

Economical

Gas on-demand, no cylinders
to change or supply stocks
to maintain

Consistent gas quality andsupply, no impurities or
running out of gas

Eliminate ongoing costs ofcylinders, manage lifetime
running costs

Green Solution
While the economic benefits of investing in a gas generator are easy to appreciate (especiallyfrom the point of view of the lab manager!), there is also a genuine environmental benefit.
Consider the carbon footprint of a pressurized cylinder of gas, delivered from depot to yourlaboratory then collected again for refilling before resuming the cycle. Depending on
geography, that could be substantial transportation distances, not to mention the energy
consumption involved in industrial gas manufacture and processing. Engineered with energy
efficiency in mind, a PEAK Scientific gas generator offers a far more environmentally
sustainable source of laboratory gas during its lifetime.

Safe

Green

Protected

No pressurized compressedgas cylinders in your lab

No repeated gas deliveries,
energy efficientComprehensive on site
warranty & service plans72.0 Horizen

One small footprint.
One giant leap.
The most advanced nitrogen generator on the market today, utilising new technologies tobring your single quad LC-MS the reliable, consistent and high purity nitrogen you need toperform routine and non-routine analyses, day after day

See the full range
www.peakscientific.com/horizen

“The size is small andpractical to keep in thelab without sophisticatedinstallations.
The operation of the Horizen
24 does not require excessive
energy consumption andneither does it require theuse of air conditioning for its
proper functioning.”
Nancy Lara Almazán
National Institute for Nuclear Research, Mexico
Horizen 24 user

One small footprint, one giant leap
Bringing best-in-class purity andunparalleled energy-efficiency together
in an ultra-compact footprint, Horizen 24
heralds a giant leap forward in nitrogengeneration for your lab. Giving you theperfect combination of performance andeconomy while helping to reduce yourlab’s carbon footprint.

 Save space with a 28% smaller footprint*
 Save money with reduced operating
costs, lower heat output and minimal
downtime
 Save the planet with an energy
consumption reduction of up to 55%***compared to existing equivalent PEAK model
10**benchmarked against equivalent competitor

Horizen

Horizen 24
Bringing Single Quad LC-MS users the very latest in nitrogen gas generator technology, Horizen 24 is theculmination of over 25 years at the forefront of nitrogen gas generation for the lab from Peak Scientific.
Designed to provide a premium, yet cost-effective nitrogen solution, Horizen 24 has incorporated state of the art
compressor technology to efficiently manage power consumption reducing energy costs and heat output withoutcompromising on performance.
As a standalone solution delivering 24 liters per minute of high purity nitrogen, up to 99.9%, from a compact
footprint, Horizen 24 provides labs peace of mind and confidence in their LC-MS results.
Features
The most energy-efficient nitrogen generator on
the market
Up to 24LPM of ultra- dry, non-methanehydrocarbon free nitrogen at up to 99.9% purity
and 116psi
55% less energy consumption, saving on cost of
power*>50% less heat, reducing your air conditioning cost*
The smallest nitrogen generator in its class, fitting
easily under any lab bench
Minimized operational carbon footprint comparedwith cylinders and equivalent generator models
Convenient fixed annual service interval on all parts
Collision Gas Add-On is compatible withthis generator

Product

Flow Rate

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Horizen 24

Up to 24 L/min

Nitrogen100-116 psi /524 x 450 x 719 mm

LC-MS, ELSD, Small sample

CE, CSA, FCC

6.8-8 bar

20.6 x 17.7 x 28.3 ”

evaporators* Benchmarked against equivalent competitor

113.0 Genius

It’s good to know you havea Genius in your lab
Our Genius series is the culmination of over a decade’s work in perfecting nitrogen gas
generators for LC-MS & LC-MS/MS applications. These generators deliver greater efficiency,superior reliability and improved performance than their predecessors.

See the full range
www.peakscientific.com/genius

Product

Flow Rate

Gas Output

OutputPressure

Size (HxWxD)

Applications

Accreditations

Genius SQ 24

Up to 24 L/min

Nitrogen

Up to 116 psi / 8 bar

610 x 600 x 750 mm

Single Quad LC-MS

CSA,CE, FCC

LC-MS/MS

CSA, CE, FCC

LC-MS/MS

CSA, CE, FCC
CSA, CE, FCC>95%
Genius XE 35

Up to 35 L/min

Nitrogen24.1 x 23.7 x 29.6’’
Up to 116 psi / 8 bar

up to 99.5%
Nitrogen

650 x 570 x 710 mm
25.6 x 22.5 x 28’’

Genius XE 70

Up to 70 L/min

Up to 116 psi / 8 bar 1000 x 570 x 710 mm

Genius XE SCI 2

26 L/minNitrogen60 psi / 4.1 bar

700 x 570 x 897 mm

Echo® MS System

32 L/minDry Air

105 psi / 7.2 bar

28 x 23 x 36”

SCIEX Triple Quad™ 7500 LC-MS/MS

25 L/minDry Air

70 psi / 4.8 bar

up to 99.5%

39.4 x 22.5 x 28’’

System
ZenoTof 7600 LC-MS/MS system

Genius XE SMZ

27 L/minNitrogen

100 psi / 6.9 bar

700 x 570 x 897 mm

LCMS-8060NX,9030 or

29 L/minAir

100 psi / 6.9 bar

27.6 x 22.5 x 35.4”

LCMS-8045,8050,8060 in Standard or HighCE, FCC

Gas Delivery mode
Genius XE QSDGenius 1053

Nitrogen80 psi / 5.5 bar

700 x 570 x 897 mm

Dual and single source PerkinElmer QSight

Air

110 psi / 7.6 bar

27.6 x 22.5 x 35.4’’

100, 200 & 400 Series

Nitrogen

100 psi / 6.9 bar

713 x 600 x 750 mm

Spectroscopy, i.e. CD Analyzers, ICP-OES

CSA, CESCIEX LC-MS/MS (excluding MD)

CSA, CE

CSA, CE, FCC

Up to 20 L/min

up to 99.995%
Genius 1024

Genius 1025
Genius 3045

CSA, CE, FCC

16 L/min
67 L/min28.1 x 23.7 x 29.6’’

19 L/minNitrogen

65 psi / 4.5 bar

713 x 600 x 750 mm

26 L/minDry Air

100 psi / 6.9 bar

28.1 x 23.7 x 29.6’’

25 L/minDry Air

60 psi / 4.1 bar

15 L/minNitrogen80 psi / 5.5 bar

713 x 600 x 750 mm

PerkinElmer QSight Triple Quad 110, 210

35 L/minDry Air

110psi / 7.6 bar

28.1 x 23.7 x 29.6’’

LCMS/MS Systems
EVOQ QQQ

CSA, CE

CSA, CE *32 L/minNitrogen80 psi / 5.5 bar

1322 x 600 x 850 mm

50 L/minDry Air

80 psi / 5.5 bar

52.1 x 23.7 x 33.5’’

Genius 1051 /Up to 25 L/min

Nitrogen

110psi / 7.6 bar

713 x 600 x 750 mm

8045, 8050 and 8060 LC-MS/MS in Default

Genius 1061

Combined

Dry Air

100 psi / 6.9 bar

28.1 x 23.7 x 29.6’’

Gas Mode* 1061 is for Japanese market and only has CE & CB accreditation.

13

Genius

Genius XE SeriesInspired by the success of our best-selling Genius line of nitrogen gas generators for LC-MS/MS, Genius XE
Nitrogen is a cutting-edge evolution combining advanced technology with refined and robust engineering. With twomodels - XE 35 (up to 35 L/min) and XE 70 (up to 70 L/min) - Genius XE Nitrogen provides a premium, standalone
nitrogen solution for high performance LC-MS/MS and other mission-critical laboratory applications where
performance and reliability are paramount.
Featuring Multi-Stage Purification™ and next-generation integrated compressors with Electronic Compressor
Optimization™ (ECO) technology, Genius XE delivers factory certifiable purity up to 99.5% on-demand, 24/7 with aconvenient fixed annual service interval.
Features
Variable flow up to 70 L/min
Variable pressure up to 116 psi
Multi-Stage Purification™ producing analyticalgrade nitrogen gas up to 99.5% purity
2 year comprehensive manufacturer’s warranty*
Touch-screen full colour user interface for easeof operation
Additional compressor capacity (Genius XE 70 only)
Collision Gas Add-On is compatible withthis generator* Year 2 of warranty subject to generator being serviced at end of year 1 by a PEAK-approved agent in accordance with fixed annual maintenance schedule.
For full terms and conditions visit www.peakscientific.com/warranty-statement

Product

Flow Rate

Genius XE 35

up to 35 L/min

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Nitrogen

116 psi / 8 bar

640 x 570 x 710 mm

LC-MS/MSCE, CSA, FCC

LC-MS/MSCE, CSA, FCC

25.2 x 22.5 x 28’’

up to 99.5%
Genius XE 70

up to 70 L/min

Nitrogenup to 99.5%

14

116 psi / 8 bar

1000 x 570 x 710 mm
39.4 x 22.5 x 28’’

Genius

Genius SQ 24
The Genius SQ 24 nitrogen gas generator provides a dedicated supply of laboratory-grade nitrogen for labs
utilizing any Single Quad LC-MS systems. With a compact size to fit under most lab benches and flow rates of up
to 24 L/min, exceeding the maximum flow rate required by any Single Quad LC-MS on the market today, your lab
can enjoy hassle-free nitrogen gas which helps you deliver the best analytical results.

FeaturesBest value nitrogen generator on the market
Variable flow up to 24 L/min
Laboratory grade nitrogen supply for Single
Quad LC-MS
Outlet pressure up to 116psi (at 22 L/min maximum)
Reduced height and compact size to fit undermost lab benches
Integrated internal compressors for easy plugand play

Product

Flow Rate

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Genius SQ 24

Up to 24 L/min

Nitrogen

Up to 116 psi / 8 bar

610 x 600 x 750 mm

Single Quad LC-MS

CSA,CE, FCC>95%

24.1 x 23.7 x 29.6’’

15

Genius

Genius 1053
Producing up to 20 L/min of high purity nitrogen (99.995%), the Genius 1053 is suitable for a range ofSpectroscopy applications, such as Circular Dichroism. Developed using (Pressure Swing Adsorption) PSA
technology, this generator produces variable purities depending on outlet flow.
Genius series gas generators offer a self-sufficient, continuous supply of laboratory-grade nitrogen to your lab,eliminating the need for gas cylinders and external sources of house air or compressors.

Features
Innovative CMS and PSA technology ensuringhigh purity nitrogen
Compressor based solution, no need for an
external air supply
Minimal set-up required
Highly economical source of nitrogen gas withlow lifetime running costs
Gas is supplied on demand so generator works
to your schedule

Product

Flow Rate

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Genius 1053

up to 20 L/min

Nitrogen

100 psi / 6.9 bar

713 x 600 x 750 mm

Spectroscopy including CD

CSA, CE

28.1 x 23.7 x 29.6’’

Analyzers and ICP-OES

up to 99.995%

16

Genius

Genius XE SCI 2
A premium standalone nitrogen and air solution for the latest SCIEX LC-MS systems, Genius XE SCI 2 also
supports the higher gas flow requirements of the Echo MS System and ZenoTof 7600 System.
Featuring Multi-Stage Purification™ and next-generation integrated compressors with Electronic Compressor
Optimization™ (ECO) technology, Genius XE SCI 2 delivers a reliable and cost-efficient source of nitrogen(Curtain Gas™) and clean, dry, oil-free air for source and exhaust gas at flows and pressures configured to meetthe latest SCIEX LC-MS system requirements.

Features24/7 performance- next generation highperformance premium compressors, ensuring
maximum uptime of your LC-MS instrument
ECO (Electronic Compressor OptimisationTM)
technology for increased compressor durability
and reduced carbon footprint
Repeatability- Multi-Stage Purification™ to
reduce moisture and contaminants providing
consistent quality of gas
Low impact on laboratory with low noise andheat emissions
Significantly reduced footprint, to fit undermost standard lab benches
No external compressed air source required
Fixed Annual Service Interval- service downtime
limited to once a year

Product

Flow Rate

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Genius XE SCI 2

26 L/min32 L/min
25 L/minNitrogen
Dry Air
Dry Air

60 psi / 4.1 bar
105 psi / 7.2 bar
70 psi / 4.8 bar

700 x 570 x 897 mm

Echo® MS System

CSA, CE, FCC

28 x 23 x 36”

SCIEX Triple Quad™ 7500 LC-MS/MS System
ZenoTof 7600 LC-MS/MS system174.0 Solaris

See lab gas in a new lightWith flow rates up to 35 L/min, Solaris generators are compact and economical nitrogengas solutions for labs.

See the full range

www.peakscientific.com/nitrogen-generators/solaris

“It is small, looks nice andis very quiet. The shapemakes it very easy to place
underneath our desk.”
Maria Fedorova
Centre for Biotechnology and Biomedicine
at University of Leipzig, Germany

Solaris

Solaris XE
PEAK Scientific’s Solaris XE nitrogen generator has been engineered using membrane technology to meet the gasdelivery requirements for labs using LC-MS/MS or multiple ELSDs. Solaris XE can operate with varying flow rates,
purity and outlet pressure, offering a flexible solution for a variety of applications.
The Solaris XE has been engineered to provide nitrogen to laboratories that utilize an external source of compressed
air, which meets a minimum quality grade of ISO 8573-1:2010 Class 1.4.1. Its compact, space-saving chassis allows it
to be placed on a benchtop, mounted on a wall or beneath an instrument, taking up minimal space in the lab.

FeaturesNitrogen up to 35L/min, up to 116 psi and up to
99.5% purity
Variable outlet flow, pressure and purity
Compact, stackable system maximizing the useof valuable laboratory floor or bench space
Colour indicative LED lighting for easy statusawareness
Manufactured and tested to highest spec in UK,
CE/FCC/CSA certified
Collision Gas Add-On is compatible withthis generator

Product

Flow Rate

Solaris XE

Up to 35 L/min

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Nitrogen

Up to 116 psi / 8 bar

156 x 343 x 650 mm

LC-MS, LC-MS/MS or

CE, CSA, FCC

6.2 x 13.5 x 25.6’’

Multiple ELSDs

up to 99.5%

20

Solaris

Solaris 1010A
PEAK Scientific’s Solaris 1010A nitrogen generator has been engineered using membrane technology to meet thenitrogen gas requirements for labs using ELSD with their High Performance Liquid Chromatography, Compact
Mass Spectrometer or for TLD reader instruments. Solaris 1010A will deliver flow rates of up to 10L/ min and a gas
purity of up to 99.5%. The outlet pressure is adjustable to 100psi max offering a flexible solution for labs.

FeaturesNitrogen up to 10 L/min at up to 100 psi with apurity of up to 99.5% at 4L/min
Variable outlet flow, pressure and purity
Colour indicative LED lighting for easy statusawareness
Manufactured and tested to highest spec in UK,
CE/ FCC certified
Safer and more convenient than pressurized
cylinders, dewars or bulk storage
12 month on-site comprehensive warranty

Product

Flow Rate

Solaris 1010A

up to 10 L/min

Gas Output

Output Pressure

Size (HxWxD)

Nitrogen

100 psi / 6.9 bar

up to 99.5%

Applications

Accreditations

453 x 417 x 540 mm

ELSD, Compact Mass

CE, CSA, FCC

17.8 x 16.4 x 21.3 ”

Spectrometers or TLD
Readers21

5.0 Infinity

Compressor-free N2generators for your lab
Infinity generators are able to run uninterrupted 24 hours a day. Effectively silent in
operation these generators can deliver up to 500 L/min of high purity nitrogen.

See the full range
www.peakscientific.com/infinity

Output
Pressure

Size (HxWxD)

Applications

Accreditations

Nitrogen
Dry Air

135 Psi (Depending

995 x 410 x 950 mm

For multiple lab instruments requiring

CE, CSAon Inlet Pressure)339.2 x 16.1 x 37.4”

nitrogen or air

Nitrogen
Dry Air

135 Psi (Depending

995 x 410 x 950 mm

For multiple lab instruments requiring

on Inlet Pressure)339.2 x 16.1 x 37.4”

nitrogen or air

Nitrogen
Dry Air

135 Psi (Depending

995 x 410 x 950 mm

For multiple lab instruments requiring

up to 210 L/min

on Inlet Pressure)339.2 x 16.1 x 37.4”

nitrogen or air

Infinity XE 6040 up to 500 L/min

Nitrogen

135 Psi (Depending

995 x 410 x 950 mm

For multiple lab instruments requiring

on Inlet Pressure)339.2 x 16.1 x 37.4”

nitrogen or air

Product

Flow Rate

Gas Output

Infinity XE 6010

up to 130 L/min
up to 210 L/min

Infinity XE 6020 up to 260 L/min
up to 210 L/min
Infinity XE 6030 up to 390 L/min

up to 210 L/min

CE, CSA
CE, CSA
CE, CSA

Infinity 1031

19 L/min
26 L/min
25 L/minNitrogen
Dry Air
Dry Air

65 psi / 4.5 bar
110 psi / 7.6 bar
60 psi / 4.1 bar

760 x 365 x 200 mm
30 x 14.4 x 7.9’’

1 x SCIEX LC-MS/MSCE

Infinity 1032

38 L/min
52 L/min
50 L/minNitrogen
Dry Air
Dry Air

65 psi / 4.5 bar
110 psi / 7.6 bar
60 psi / 4.1 bar

730 x 423 x 250 mm
28.8 x 16.7 x 9.9’’

2 x SCIEX LC-MS/MSCE

Infinity 1045

32 L/min
50 L/minNitrogen
Dry Air

90 psi / 6.2 bar
110 psi / 7.6 bar

730 x 424 x 250 mm
28.8 x 16.7 x 9.9’’

EVOQ QQQ

CEInfinity 1046

20 L/min
70 L/minNitrogen
Dry Air

80 psi / 5.5 bar
110 psi / 7.6 bar

730 x 421.6 x 250 mm
28.8 x 16.6 x 9.9’’

Nitrogen & Air Generator for PerkinElmer
QSight Triple Quad

CE

Infinity 1051

25 L/min(Combined
total)

Nitrogen
Dry Air

100 psi / 6.9 bar

730 x 424 x 250 mm
28.8 x 16.7 x 9.9’’

LCMS-8045/ 8050/ 8060 instruments

CE* Must be paired with a compressed air source which, as a minimum, meets ISO 8573-1:2010 Class 1.4.1

23

Infinity

Infinity XE 60 Series
Producing between 10 and 500 L/min of high purity nitrogen gas, the Infinity XE 60 series nitrogen membrane gas
generators can comfortably supply multiple laboratory instruments with instrument grade nitrogen.
To operate, the Infinity series of generators require an external source of compressed air, which meets aminimum quality grade of ISO 8573-1:2010 Class 1.4.1. Where required, PEAK can provide assistance on external
compressors & pre/post filtration as part of delivering a complete solution for your needs.

Features
Flow rates ranging from 10-500 L/min*N2 purity up to 99.5%
In-field expansion of nitrogen capacity to meetgrowing lab demands
Optional second outlet for Dry Air
IMM reduces overall compressor run time andenergy consumption, reducing carbon footprint
Fully comprehensive 2-Year Warranty**
Collision Gas Add-On is compatible with thisgenerator* Dependent on inlet pressure, inlet/outlet flow & Nitrogen purity setting** subject to meeting 2 year warranty conditions www.peakscientific.com/
warranty-statement/

24

Product

Flow Rate

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Infinity XE 60XX

up to 500 L/min

Nitrogen

135 Psi (Depending

995 x 410 x 950 mm

For multiple lab instruments

CE, CSA

up to 210 L/min

Air

on Inlet Pressure)339.2 x 16.1 x 37.4”

requiring nitrogen or air

Infinity

Infinity 1031
The Infinity 1031 delivers high purity nitrogen and dry air to support SCIEX LC-MS systems.
A compressor-free solution, the Infinity 1031 requires an air source to operate. With few moving parts thisgenerator has minimal maintenance requirements and can operate 24 hours a day for hassle-free nitrogen.

Features
Capable of supplying most SCIEX LC-MS
systems
Exhaust, curtain and source gases from a single
generator
24/7 operation at optimum performance if
requiredCompletely silent in operation

Product

Flow Rate

Gas Output

Output Pressure

Size (HxWxD)

Applications

Accreditations

Infinity 1031

19 L/min
26 L/min
25 L/minNitrogen
Dry Air
Dry Air

65 psi / 4.5 bar
110 psi / 7.6 bar
60 psi / 4.1 bar

760 x 365 x 200 mm
30 x 14.4 x 7.9’’

1 x SCIEX LC-MS/MS

CE

25

6.0 Precision

Streamline your GC
workflow with Precision
Combining convenience and reliability in a stackable and modular design, Precision is thesafe and practical GC gas solution.

See the full range

www.peakscientific.com/precisionProduct

Flow Rate

Gas Output

Precision Hydrogen
SL 100100 cc/min

Hydrogen99.9995%

Precision Hydrogen
SL 200200 cc/min

Precision Hydrogen
Trace 250

OutputPressure

Size (HxWxD)

Applications

Accreditations

100 psi / 6.9 bar

250 x 160 x 355 mm
9.9 x 6.3 x 14’’

GC flame detectors

CSA, CE

Hydrogen99.9995%

100 psi / 6.9 bar

250 x 160 x 355 mm
9.9 x 6.3 x 14’’

GC flame detectors

CSA, CE

250 cc/min

UHP Hydrogen
99.99999%*

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC & GC-MS Carrier & Detector
Gas, ICP-MS Reaction gas

CSA, CE

Precision Hydrogen
Trace 500

500 cc/min

UHP Hydrogen
99.99999%*

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC & GC-MS Carrier & Detector
Gas, ICP-MS Reaction Gas

CSA, CE

Precision Hydrogen
Trace 1200

1200 cc/min

UHP Hydrogen
99.99999%*

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC & GC-MS Carrier & Detector
Gas, ICP-MS Reaction Gas

CSA,CE

Precision Hydrogen100

100 cc/min

Hydrogen99.9995%

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas, ICP-MS Reaction
Gas

CSA, CE

Precision Hydrogen200

200 cc/min

Hydrogen99.9995%

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas, ICP-MS Reaction
Gas

CSA, CE

Precision Hydrogen300

300 cc/min

Hydrogen99.9995%

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas, ICP-MS Reaction
Gas

CSA, CE

Precision Hydrogen450

450 cc/min

Hydrogen99.9995%

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas, ICP-MS Reaction
Gas

CSA, CE

Precision Hydrogen
1200

1200 cc/min

Hydrogen99.9995%

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas, ICP-MS Reaction
Gas

CSA, CE

Precision Nitrogen
Trace 250

250 cc/min

Zero Nitrogen
99.9995%

80 psi / 5.5 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC & GC-MS Carrier & Detector Gas

CSA, CE

Precision Nitrogen
Trace 600

600 cc/min

Zero Nitrogen
99.9995%

80 psi / 5.5 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC & GC-MS Carrier & Detector Gas

CSA, CE

Precision Nitrogen
Trace 1L

1000 cc/min

Zero Nitrogen
99.9995%

80 psi / 5.5 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC & GC-MS Carrier & Detector Gas

CSA, CE

Precision Nitrogen250

250 cc/min

UHP Nitrogen
99.9995%

80 psi / 5.5 bar

256 x 380 x 540 mm
10.1 x 15 x 21.3’’

GC Detector Gas

CSA, CE

Precision Nitrogen
Headspace 250

250 cc/min

UHP Nitrogen
99.9995%

100 psi / 6.9 bar

256 x 380 x 540 mm
10.1 x 15 x 21.3’’

GC Sample Preparation and
Detector Gas

CE

Precision Nitrogen600

600 cc/min

UHP Nitrogen
99.9995%

80 psi / 5.5 bar

256 x 380 x 540 mm
10.1 x 15 x 21.3’’

GC Detector Gas

CSA, CE

Precision Nitrogen
1L

1000 cc/min

UHP Nitrogen
99.9995%

80 psi / 5.5 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas

CSA, CE

Precision Zero Air
1.5

1.5 L/min

Zero Air<0.05ppm

80 psi / 5.5 bar

156 x 380 x 540 mm
6.2 x 15 x 21.3’’

GC Detector Gas

CSA, CE

Precision Zero Air
3.5L

3.5 L/min

Zero Air<0.05ppm

80 psi / 5.5 bar

156 x 380 x 540 mm
6.2 x 15 x 21.3’’

GC Detector Gas

CSA, CE

Precision Zero Air 7

7 L/minZero Air<0.05ppm

80 psi / 5.5 bar

256 x 380 x 540 mm
10.1 x 15 x 21.3’’

GC Detector Gas

CE

Precision Zero Air
18L

18 L/minZero Air<0.05ppm

80 psi / 5.5 bar

256 x 380 x 540 mm
10.1 x 15 x 21.3’’

GC Detector Gas

CE

Precision Zero Air
30L

30 L/minZero Air<0.05ppm

100 psi / 6.9 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

GC Detector Gas

CSA, CE

Precision Air
Compressor-

Compressed Air

120 psi / 8.3 bar

406 x 380 x 540 mm
16 x 15 x 21.3’’

Independent air supply for Precision
modules

CSA, CE* Based on O2 content independently verified by National Physical Laboratory, UK

27

The smallest andeasiest to use hydrogengenerator for GC-FID.
Precision Hydrogen SL is everything we know
about hydrogen generation: streamlined toperfection for GC-FID. We’ve designed it to besmaller (less than a quarter of the size of its
predecessor), child’s play to operate and maintainand, above all, a safer alternative to having
compressed cylinders in your lab for GC flame
detector gas.

Small
Available in both 100 and 200cc models,
Precision SL is the smallest laboratoryhydrogen generator in its class, minimizing thelaboratory space required for GC detector gas.

SimpleProduce hydrogen gas at the push of a button.
With only simple user maintenance required (under
60 seconds), look forward to constant GC flame
detector gas.

Safe
Uncompromised safety in your laboratory. Unlikepressurized cylinders, gas is generated on-demandwith minimal gas stored plus advanced fail-safe
technology.

{image almost to scale}

Available in black or white and with two models to choose from - 100cc and 200cc.

Precision

Precision SL
The smallest hydrogen generator for GC-FID, the Precision Hydrogen SL line has been developed to offer a
streamlined laboratory-grade hydrogen gas solution for GC flame detectors at a purity of 99.9995%. Precision
Hydrogen SL is safe, simple to use, easy to maintain and takes up minimal bench space.
With an impressive, streamlined form factor, Precision Hydrogen SL produces hydrogen at up to 200cc/min withno compromise on safety. A range of features ensure that Precision Hydrogen SL will produce hydrogen safely,
storing a minimal volume of gas compared to cylinders.
This generator comes in 2 models delivering hydrogen at up to 100cc/min and 200cc/min, a choice of black orwhite and includes a 2 year warranty as standard.
Features2 models – 100cc and 200cc
99.9995% purity @ 100 psi
One-button start-up and shutdown
Easy end-user managed servicing
Optional auto-water fill
Auto-shutdown fail-safe feature
CE, CSA compliant
2 year comprehensive warranty*
Available in black or white* Conditional on replacement of DI cartridge after 12 months.

Need help calculating your GC gas flows?
Visit peakscientific.com/gasflow

29

Precision

Precision Hydrogen
The Precision Hydrogen generators are designed to provide high quality gas needed for GC detectors requiringhydrogen fuel gas, such as FID. The Precision Hydrogen Trace generators are also capable of supplying GC carrier
gas as well as detector gases for GC and GC/MS. One generator is capable of supplying multiple GCs and GC
detectors with various flow rates available to suit individual laboratory needs. These generators utilize a Proton
Exchange Membrane (PEM) to create hydrogen gas from deionized water and are equipped with a dryer toprovide high purity hydrogen. The Precision Hydrogen generators are also suitable for providing collision gas for
ICP-MS, whilst Hydrogen Trace is also suitable for reaction gas.
Precision Hydrogen gas generators come with various safety features as standard, giving you complete peace ofmind in the laboratory and are a far safer, dependable and convenient alternative to cylinders.
Features99.9995% purity (standard) / 99.99999% purity(trace)Suitable for GC detector gas (all) and carriergas (trace model only)
Creates hydrogen on demand, minimal storageof hydrogen in the system
Low maintenance and minimal running costs
over product lifetime
Internal leak detection with automatic shutdown
Automatic loading pump as standard

Need help calculating your GC gas flows?
Visit peakscientific.com/gasflow

Precision

Precision Nitrogen
The Precision Nitrogen generators are available in both standard and trace models. The standard models comein three flow rate models (250cc, 600cc and 1000cc) which are able to provide make-up gas, reference gas andgas for nitrogen sample preparation (i.e. headspace). The Nitrogen Trace models are engineered to producehydrocarbon-free zero nitrogen as carrier gas, make-up gas and reference gas for sample preparation available in
250cc, 600cc and 1000cc models.
These generators are capable of delivering high purity nitrogen, removing oxygen and moisture via Pressure Swing
Adsorption and Carbon Molecular Sieve technology and a cat chamber is employed for Hydrocarbon removal(Hydrocarbon removed as methane - trace models only). As with all Precision Series generators, nitrogen models
benefit from a compact and modular, stackable design, minimizing the total footprint required for GC gas supply
and providing flexibility to add or remove modules as your laboratory requirements grow over time.
Features
All models offer 99.9995% purity
Nitrogen Trace model suitable as GC and GC/
MS carrier and detector gas
Trace capable of delivering hydrocarbon-free
nitrogen suitable for carrier gas, make-up gas
and sample preparation
Highly economical source of nitrogen gas withlow lifetime running costs
Ultra-fast start-up time, quick to reach standard
operating purity

Need help calculating your GC gas flows?
Visit peakscientific.com/gasflow

31Precision

Precision Zero Air
The Precision Zero Air generators are designed specifically to supply clean, dry, hydrocarbon free air to be used as
flame support gas for GC.
As with all Precision Series generators, Zero Air benefits from a compact and modular, stackable design,
minimizing the total footprint required for GC gas supply, and providing flexibility to add or remove modules as
your laboratory requirements evolve over time.

FeaturesHydrocarbon content <0.05ppm for marketleading purity
Integrates seamlessly with other Precision units
Minimum lifetime maintenance requirements, no
expensive catalyst chamber replacements
Highly economical source of dry, hydrocarbon
free air
Avoid risk of contaminants entering the system(when switching out empty cylinders)

Need help calculating your GC gas flows?
Visit peakscientific.com/gasflow

“We decided to switch tohydrogen gas generators to
reduce the costs of helium
cylinder deliveries, and tosave time by eliminating
the need to change overcylinders.”
Federico Cozzi
Laboratory Manager, Copenhagen University
Department of Plant and Environmental Sciences,
DenmarkPrecision

Modular configurations
GC flame detector gas•

Precision Zero Air•

Precision Nitrogen (option for make-up gas)•

Precision Hydrogen

GC H2 carrier gas with FID•

Precision Zero Air•

Precision Nitrogen (option for make-up gas)•

Precision Hydrogen Trace

GC-MS carrier gas•

34

Precision Hydrogen Trace

The Precision Series can be stacked in multiple variations depending on your lab’s particular
GC requirement, whether it is to supply GC carrier gas on its own, delivering flame support gas
to detectors or for other detectors like TCDs or ECDs.
The Precision Series can also be purchased with an optional, stackable air compressor module
for labs without a suitable air supply for the nitrogen or zero air generators.
For a full method list visit: www.peakscientific.com/gc-methods/

GC N2 carrier gas with FID•

Precision Zero Air•

Precision Nitrogen Trace•

Precision Hydrogen

GC-ECD or GC-TCD
carrier & detector gas•

Precision Nitrogen Trace35

7.0 Specialist Solutions

Every lab needs a specialist
Beyond our core ranges of laboratory gas generators, we have developed tailored solutions
to meet the specific demands of many key analytical applications and instrument types.

See the full range
www.peakscientific.com/lab-solutions

Product

Flow Rate

Gas Output

OutputPressure

Corona Nitrogen
1010A

5 L/min

Nitrogen

NG5000(A)*Up to 5 L/min

UHP Nitrogen

Size (HxWxD)

Applications

Accreditations

80 psi / 5.5 bar

406x417x540 mm
16x16.4x21.3”

Thermo Fisher Scientific Corona
Veo CAD, Dionex ERS/VanquishCAD

CE, CSA, FCC

80 psi / 5.5 bar

1235x430x410mm

GC, ICP, DSC, DMA,

CE

48.6x16.9x16.1”

ICP-OES660 x 472.4 x 160 mm

TC, TOC, IC, NPOC

CE, FCC, KC

FT-IR microscopes
Laboratory applications requiringCE

99.9995%

AD140L

140L/minDry Air

115psi/7.9bar

26 x 18.6 x 6.3’’

PG14L

Up to 14 L/

Dry Air

Up to 100 psi /660 x 459 x 162 mm<1ppm CO2

6.9 bar

26 x 18.1 x 6.4’’

Up to 28 L/

Dry Air

Up to 100 psi /910 x 459 x 162 mm

min<1ppm CO2

6.9 bar35.9 x 18.1 x 6.4’’

TOC 1000

6L/minDry Air

100 psi / 6.9 bar
145 psi/10.0 bar

334 x 270 x 530mm
13.5 x 10.7 x 20.9’’

TC, TOC, IC, NPOC

CE, FCC, KC

Collision Gas
Add-On**0.2L/min

UHP Nitrogen

80 psi / 5.5 bar

165 x 360 x 425mm
6.5 x 14.2 x 16.7’’

TC, TOC, IC, NPOC

CE, FCC, KC

min
PG28L

99.999%CO2 free air
FT-IR microscopes
Laboratory applications

CErequiring CO2 free air*A models include air supply** Nitrogen generator providing main gas source to an instrument. (Genius XE35/70, Infinity XE and Solaris XE systems)

8.0 i-FlowLab

Scalable high-flow, highpurity N2 solution for labs
i-FlowLab is a modular & expandable on-site nitrogen generation system. It is capableof delivering a continuous & consistent supply, at a range of purities (up to 99.9995%
N2) and flow rates to meet the full & varying gas demands of your combined laboratory
applications, such as LC-MS, gloveboxes, fume hoods, sample evaporators and more.

See the full range

www.peakscientific.com/iflowlab

Key Features)

Consistent & Convenient - Constant, reliable, stable, on-demand gas
supply that eliminates the inconvenience of changing cylinders or dewars.

)

Economical & Sustainable - A cost effective total laboratory nitrogen gas
supply solution that eliminates the need for bulk delivery.

)

Expandable & Scalable - i-FlowLab has the capacity to meet and exceed yourcurrent gas demands with the ability to expand as your laboratory grows.

)

Energy Efficient - An innovative ‘Eco-mode’ ensures the lowest runningcosts by automatically managing production based on your daily demands.

)

Safe Supply - Eliminate the handling of cylinders or storage of highlypressurized gases.

)

Verified Compliance - Exceeds standards of EIGA, EC Food Grade,
European Pharmacopoeia, JECFA and US Food & Drug Administration(CFR Title 21) . PEAK IQ/OQ certification also available.

)

High Quality Engineering - PEAK is an ISO 9001 certified manufacturer
and i-FlowLab is expertly engineered to ensure performance and reliability.

i-FlowLab

i-FlowLabi-FlowLab from PEAK Scientific provides a total laboratory solution for on-site generation of nitrogen gas,delivering a continuous and consistent supply of high-purity nitrogen at the required pressure and flow rates tomeet the full and varying demands of your laboratory or research facility.
Engineered around PSA technology, i-FlowLab is available in various pre-configured specifications to suit specificflow and purity demands. A single i-FlowLab generator installation can provide nitrogen at flow rates from
11-5590 L/min. Purities are specified at time of system design to meet the needs of the application up to 99.999%.
Thanks to the expandable design additional CMS columns can be added to each i-FlowLab generator after
installation to increase the maximum flow rate.
Features
Consistent, constant, reliable, on-demand
supply of on-site gas
No instrument or application downtime as aresult of running out of stored gas
Bring control of your nitrogen supply in-house
No more administration costs
Scalable/expandable, increase nitrogen production
Economical, fast return on investment and low
cost of ownership with predictable running cost

OutputPressure

Size (HxWxD)

Applications

Accreditations

95% - 99.9995%

87- 145 psi

1738 x 500 x 760-2200 mm

LC-MS, Fume Hoods,

CE

UHP Nitrogen/ 6 - 10 bar

68.5 x 19.7 x 30 - 86.7”

Glovebox, Sample

95% - 99.9995%

87- 145 psi

853 x 418 x 830 - 1316mm

LC-MS, Fume Hoods,

UHP Nitrogen/ 6 - 10 bar

16.5 x 33.6 x 32.7 - 51.8 ”

Glovebox, Sample

Product

Flow Rate

Gas Output

i-FlowLab

11-5590 L/min

701X-710X

Evaporators & more
i-FlowLab Mini
701X-704X

40

13-584 L/min

Evaporators & moreCE

Scalable
Up to ten CMS column banks can be added to eachsingle i-FlowLab unit in less than a day, increasing
nitrogen production capacity with minimal downtime.

ModularMultiple units can be synchronized to meet demands basedon application flow-rates and purity requirements (moreunits = greater flow-rates at specified purity).
95 - 99.9995% N2 purity, at 11-5590 L/min

Example System
Receiver tankisolation valve

Air compressor

Air compressor

Air
receiver tank

Bulk water/
moisture filter

Bulk water/
moisture filter

Air
receiver tankDryer

ISO 8573 - 1:2010
Class 2.4.1 Air Quality(Critical to meeting the
optimal standard of
the feed air.)

Dryer

Absorption
tower

Receivisolati

ISO 8573 - 1:2010
Class 2.4.1 Air Quality(Critical to meeting the
optimal standard of
the feed air.)Absorption
tower

Receiver tanki-Flow Prime N2 i-Flow Prime N2 purge ventReceiver tankpurge vent41

8.0 PEAK Protected

Not all warranties are equal
What differentiates PEAK from other gas generator manufacturers is that a PEAK gas generator
comes with a truly comprehensive on-site warranty. This means that in the unlikely event thatyour gas generator develops a problem, we will send a PEAK certified Field Service Engineer toyour laboratory to try to resolve the issue on-site and get you back up and running with minimal
fuss and disruption.
What’s more, when we say fully-comprehensive, we mean it. Every part of your generator
is covered during warranty. While other manufacturer warranties may only cover certain
components or require the return of your generator to the factory for repair, leaving you without
gas for weeks or even months, our warranty is designed to give you complete peace of mind.

Safeguard your laboratory
workflow with
Servicing any equipment in your lab should not be seen as just an option, it should be treatedas a necessity. With many moving parts and wear & tear, especially where generators feature
integrated compressors, as well as filters and other ‘consumables’, a gas generator needs to be
maintained regularly to continue to perform at its best for many years.
Here at PEAK we see your gas generator as the beating heart of your laboratory, enabling youranalytical instruments to perform and achieve the results you need. With this being the case it isextremely important you look after your ‘beating heart’ to ensure the best possible results.

42

We offer a range of [PEAK Protected]
services to suit your particular needs.

Rapid Response

Service Plans

Replacement Parts

Complete maintenance care with 24 hour
rapid response breakdown coverComplete maintenance care withguaranteed response timebreakdown cover

Genuine PEAK parts with expressdelivery, ensuring optimal performanceand lifetime

Installation

IQ/OQ

Technical Support Hotline

Trained PEAK FSE will visit to install andcommission your generator

Certified assurance for applications
requiring documented qualification

Around the clock support by phone oronline with our global technical helpdesk

Find out more about PEAK generator service plans at www.peakscientific.com/protected
43

Contact us today to discover more!
North America				Europe
Tel: +1 866 647 1649			
Tel: +44 141 812 8100
China						India
Tel: +86 21 5079 1190			
Tel: 1800 2700 946
Web: www.peakscientific.com/products

Email: discover@peakscientific.com

Copyright 2024 Peak Scientific - 2020 Product Catalogue Rev 23 RSID 465 PN 00-4070 EN - 09/10/2024