Data Sheet | 03.00

Test & Measurement

R&S®ESCI/ESCI7EMI Test ReceiverSpecifications

Version 03.00, June 2009

Specifications
Specifications apply under the following conditions: 15 minutes warm-up time at ambient temperature, specified environmental
conditions met, calibration cycle adhered to, and all internal automatic adjustments performed. Data without tolerances: typical values
only. Data designated 'nominal' applies to design parameters and is not assured by Rohde & Schwarz.

Frequency®

R&S ESCI
DC, AC coupled®
R&S ESCI7
DC coupled
AC coupled

Frequency range

9 kHz to 3 GHz
9 kHz to 7 GHz
1 MHz to 7 GHz
0.01 Hz

Resolution
Internal reference frequency (nominal)
Aging per year
Temperature drift
Internal reference frequency (nominal)
Aging per year
Temperature drift
External reference frequency
Frequency display (receiver mode)
Resolution
Frequency display (analyzer mode)
Marker resolution
Max. deviation

sweep time > 3 × auto sweep time

Frequency counter resolution
Count accuracy

selectable
S/N > 25 dB

Display range of frequency axis

R&S ESCI®
R&S ESCI7

Max. deviation of display range
Spectral purity, SSB phase noise

Residual FM

standard
after 30 days of continuous operation+5 °C to +45 °C®
R&S FSP-B4 option (OCXO)
after 30 days of continuous operation+5 °C to +45 °C®

f = 500 MHz, for f > 500 MHz see diagram
100 Hz
1 kHz
10 kHz
100 kHz, span > 100 kHz
1 MHz, span > 100 kHz
10 MHzf = 500 MHz, RBW = 1 kHz,sweep time = 100 ms–6

1 × 10–6
1 × 10–7

1 × 10–8
1 × 1010 MHznumeric display
0.1 Hzwith marker or frequency counter
span/500±(marker frequency × reference frequency
error + 0.5 % × span + 10 % × resolution
bandwidth + ½ (last digit))
0.1 Hz to 10 kHz± (marker frequency × reference
frequency error + ½ (last digit))
0 Hz, 10 Hz to 3 GHz
0 Hz, 10 Hz to 7 GHz
0.1 %< –84 dBc (1 Hz), typ. –90 dBc (1 Hz)< –100 dBc (1 Hz), typ. –108 dBc (1 Hz)< –106 dBc (1 Hz), typ. –113 dBc (1 Hz)< –110 dBc (1 Hz), typ. –113 dBc (1 Hz)< –120 dBc (1 Hz), typ. –125 dBc (1 Hz)
typ. –145 dBc (1 Hz)
typ. 3 Hz-60

SSB phase noise (dBc(1Hz))

-70
-80

center frequency 500 MHz
center frequency 3 GHz-90

center frequency 7 GHz
-100
-110
-120
-130
100 Hz

1 kHz

10 kHz

100 kHz

1 MHz

Typical phase noise at different center frequencies

2®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver

Version 03.00, June 2009

Scan (receiver mode)
Scan
Measurement time per frequency

scan of max. 10 subranges with different,
independent settings
33 μs to 100 s

selectable

Sweep (analyzer mode)
Sweep timein time domain, span = 0 Hz
resolution
in frequency domain, span ≥ 10 Hz

Max. deviation of sweep time

1 μs to 16000 s
125 ns
2.5 ms to 16000 s
1%

Resolution bandwidths
Sweep filters
3 dB bandwidths
Bandwidth accuracy≤ 100 kHz
300 kHz to 3 MHz≤ 100 kHz
300 kHz to 3 MHz
6 dB bandwidths
pulse bandwidth≤ 120 kHz
1 MHz
≤ 120 kHz
1 MHz

10 Hz to 3 MHz, in steps of 1/3/10<3%< 10 %<5< 15
200 Hz, 9 kHz, 120 kHz
1 MHz
<3%
< 10 %<5< 15

Video bandwidths

analyzer mode

1 Hz to 10 MHz, in steps of 1/3/10

FFT filters
3 dB bandwidths
Bandwidth accuracy
Shape factor 60 dB:3 dB

analyzer mode

Shape factor 60 dB:3 dB
EMI bandwidths
Bandwidth accuracy
Shape factor 60 dB:6 dB

1 Hz to 30 kHz, in steps of 1/3/10
5 %, nominal
2.5, nominal

Channel filters
Bandwidths

100/200/300/500 Hz;
1/1.5/2/2.4/2.7/3/3.4/4/4.5/5/6/8.5/9/10/
12.5/14/15/16/18 (RRC)/20/21/24.3 (RRC)/
25/30/50/100/150/192/200/300/500 kHz
1/1.228/1.28 (RRC)/1.5/2/3/3.84 (RRC)/
4.096 (RRC)/ 5 MHz(RRC = root raised cosine)Preselection
Preselection

can be switched off in analyzer mode

Bandwidths (–6 dB), nominal

R&S ESCI, R&S ESCI7< 150 kHz150 kHz to 2 MHz
2 MHz to 8 MHz
8 MHz to 30 MHz
30 MHz to 70 MHz
70 MHz to 150 MHz
150 MHz to 300 MHz
300 MHz to 600 MHz
600 MHz to 1 GHz
1 GHz to 2 GHz
2 GHz to 3 GHz®
R&S ESCI7
3 GHz to 7 GHz
switchable, between preselection and1st mixer

Preamplifier®

®

R&S ESCI: 11 preselection filters®
R&S ESCI7: 12 preselection filters®

230 kHz, fixed-tuned lowpass filter
2.6 MHz, fixed-tuned bandpass filter
2 MHz, tracking bandpass filter
6 MHz, tracking bandpass filter
15 MHz, tracking bandpass filter
30 MHz, tracking bandpass filter
60 MHz, tracking bandpass filter
80 MHz, tracking bandpass filter
100 MHz, tracking bandpass filter
tracking highpass filter
fixed-tuned highpass filter
tracking bandpass filter
20 dB

®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver3

Version 03.00, June 2009

Level
displayed average noise level (DANL)
to 30 dBm

Display range
Maximum input level
DC voltage

DC-coupled
AC-coupled
RF attenuation 0 dB
RF attenuation ≥ 10 dB
RF attenuation 0 dB
RF attenuation ≥ 10 dB, 10 μs®
R&S ESCI
RF attenuation ≥ 10 dB, 20 μs®
R&S ESCI7
RF attenuation ≥ 10 dB, 10 μs

CW RF power
Pulse spectral density
Max. pulse voltage
Max. pulse energy

Intermodulation
1 dB compression of input mixer
Third-order intercept (TOI)

Second harmonic intercept (SHI)

4

®

0V
50 V
20 dBm30 dBm
97 dBμV/MHz
150 V
10 mWs
1 mWs

f > 200 MHz, RF attenuation 0 dB,
5 dBm, nominal
preselection and preamplifier off
RF attenuation 0 dB, level 2 × –30 dBm, ∆f > 5 × RBW or 10 kHz, whichever is larger
without preselection, without preamplifier®
®
R&S ESCI, R&S ESCI7
20 MHz to 200 MHz> 5 dBm
200 MHz to 3 GHz> 7 dBm, typ. 10 dBm®
R&S ESCI7
3 GHz to 7 GHz> 10 dBm, typ. 15 dBmwith preselection, without preamplifier®
®
R&S ESCI, R&S ESCI7
20 MHz to 200 MHz> 0 dBm
200 MHz to 3 GHz> 2 dBm, typ. 5 dBm®
R&S ESCI7
3 GHz to 7 GHz> 10 dBm, typ. 15 dBmwith preselection, with preamplifier®
®
R&S ESCI, R&S ESCI7
20 MHz to 200 MHz> –20 dBm
200 MHz to 3 GHz> –18 dBm, typ. –15 dBm®
R&S ESCI7
3 GHz to 7 GHz> –10 dBm, typ. –5 dBm
RF attenuation 0 dB, level –10 dBm, without preselection, without preamplifier®
®
R&S ESCI, R&S ESCI7< 100 MHztyp. 25 dBm
100 MHz to 1.5 GHz
typ. 35 dBm®
R&S ESCI7
1.5 GHz to 3.5 GHz
typ. 70 dBm
RF attenuation 0 dB, level –15 dBm, with preselection, without preamplifier®
®
R&S ESCI, R&S ESCI74 MHz to 100 MHz> 40 dBm
100 MHz to 1.5 GHz> 50 dBm®
R&S ESCI7
1.5 GHz to 3.5 GHz
typ. 70 dBm
RF attenuation 0 dB, level –35 dBm, with preselection, with preamplifier®
®
R&S ESCI, R&S ESCI74 MHz to 100 MHz> 25 dBm
100 MHz to 1.5 GHz> 35 dBm®
R&S ESCI7
1.5 GHz to 3.5 GHz
typ. 10 dBm

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver

Version 03.00, June 2009

Displayed average noise level (DANL)(analyzer mode)

RF attenuation 0 dB, RBW = 10 Hz,
VBW = 1 Hz, span = 0 Hz, trace average function over 20 sweeps, 50 Ω termination
without preselection, without preamplifier, AC-coupled®
R&S ESCI
9 kHz< –105 dBm, nominal
100 kHz< –110 dBm, nominal
1 MHz< –130 dBm, nominal
10 MHz to 1 GHz< –142 dBm, typ. –145 dBm
1 GHz to 2.5 GHz< –140 dBm, typ. –143 dBm
2.5 GHz to 3 GHz< –138 dBm, typ. –141 dBm®
R&S ESCI7
1 MHz< –128 dBm, nominal
10 MHz to 1 GHz< –140 dBm, typ. –143 dBm
1 GHz to 2.5 GHz< –138 dBm, typ. –141 dBm
2.5 GHz to 3 GHz< –136 dBm, typ. –139 dBm
3 GHz to 7 GHz< –138 dBm, typ. –141 dBm
without preselection, without preamplifier, DC-coupled®
R&S ESCI
9 kHz< –115 dBm
100 kHz< –120 dBm
1 MHz< –140 dBm, typ. –143 dBm
10 MHz to 1 GHz< –142 dBm, typ. –145 dBm
1 GHz to 2.5 GHz< –140 dBm, typ. –143 dBm
2.5 GHz to 3 GHz< –138 dBm, typ. –141 dBm®
R&S ESCI7
9 kHz< –115 dBm
100 kHz< –120 dBm
1 MHz< –138 dBm, typ. –141 dBm
10 MHz to 1 GHz< –140 dBm, typ. –143 dBm
1 GHz to 2.5 GHz< –138 dBm, typ. –141 dBm
2.5 GHz to 3 GHz< –136 dBm, typ. –139 dBm
3 GHz to 7 GHz< –138 dBm, typ. –141 dBm
with preselection, without preamplifier, DC-coupled®
R&S ESCI
9 kHz< –115 dBm
100 kHz< –120 dBm, typ. –140 dBm
1 MHz< –140 dBm, typ. –148 dBm
10 MHz to 1 GHz< –142 dBm, typ. –150 dBm
1 GHz to 2.5 GHz< –140 dBm, typ. –148 dBm
2.5 GHz to 3 GHz< –138 dBm, typ. –141 dBm®
R&S ESCI7
9 kHz< –115 dBm
100 kHz< –120 dBm, typ. –140 dBm
1 MHz< –138 dBm, typ. –146 dBm
10 MHz to 1 GHz< –140 dBm, typ. –148 dBm
1 GHz to 2.5 GHz< –138 dBm, typ. –146 dBm
2.5 GHz to 3 GHz< –136 dBm, typ. –139 dBm
3 GHz to 7 GHz< –138 dBm, typ. –141 dBm
with preselection, with preamplifier, DC-coupled®
R&S ESCI
9 kHz< –135 dBm
100 kHz< –140 dBm
1 MHz< –150 dBm, typ. –153 dBm
10 MHz to 1 GHz< –152 dBm, typ. –155 dBm
1 GHz to 3 GHz< –150 dBm, typ. –153 dBm®
R&S ESCI7
9 kHz< –135 dBm
100 kHz< –140 dBm
1 MHz< –148 dBm, typ. –151 dBm
10 MHz to 1 GHz< –150 dBm, typ. –153 dBm
1 GHz to 7 GHz< –148 dBm, typ. –151 dBm®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver5

Version 03.00, June 2009

Noise indication (receiver mode)
Average (AV) display

Increase of DANL relative to AV display

Immunity to interference
Image frequency
Intermediate frequency
Spurious response

Nominal, calculated from DANL data, 0 dB RF attenuation, 50 Ω termination
without preamplifier®
®
R&S ESCI, R&S ESCI79 kHz, BW = 200 Hz< 5 dBμV
150 kHz, BW = 200 Hz< 0 dBμV
150 kHz, BW = 9 kHz< 16 dBμV
1 MHz, BW = 9 kHz< –4 dBμV
10 MHz to 30 MHz, BW = 9 kHz< –6 dBμV
30 MHz to 1 GHz, BW = 120 kHz< 6 dBμV
1 GHz to 3 GHz, BW = 1 MHz< 16 dBμV®
R&S ESCI7
3 GHz to 7 GHz, BW = 1 MHz< 20 dBμV
with preamplifier®
®
R&S ESCI, R&S ESCI79 kHz, BW = 200 Hz< –15 dBμV
150 kHz, BW = 200 Hz< –20 dBμV
150 kHz, BW = 9 kHz< –4 dBμV
1 MHz, BW = 9 kHz< –14 dBμV
10 MHz to 30 MHz, BW = 9 kHz< –16 dBμV
30 MHz to 1 GHz, BW = 120 kHz< –4 dBμV
1 GHz to 3 GHz, BW = 1 MHz< 6 dBμV®
R&S ESCI7
3 GHz to 7 GHz, BW = 1 MHz< 3 dBμV
max peaktyp. +11 dB
RMS
typ. +1 dB
quasi-peakband Atyp. +3 dBband Btyp. +4 dBbands C and D
typ. +6 dB

f > 1 MHz, 0 dB RF attenuation,
without input signal∆f > 100 kHz, mixer level < –10 dBm
field strength 3 V/m, 0 dB RF attenuation,
50 Ω termination, f ≠ fIF

Other interfering signals
RF shielding

Level display (receiver mode)
Level display

Spectrum

digitalresolution
analoglevel axis
frequency axis
Three detectors can be switched on
simultaneously.

Detectors
Units of level display
Measurement time
Level display (analyzer mode)
Screenselectable

Logarithmic level display range
Linear level display range
Number of traces

one measurement diagram
two measurement diagrams

Trace detectors
Trace functions
Number of measurement points

6®

default value
range

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver> 70 dB> 70 dB< –103 dBm< –70 dBc
level indication < 10 dBμV, nominal

numeric,
0.01 dBbargraph display separate for eachdetector
10 dB to 200 dB in steps of 10 dB
linear or logarithmic selectableaverage (AV), RMS, max peak, min peak,quasi-peak (QPK), CISPR-AV, CISPR-RMS
dBμV, dBm, dBμA, dBpW, dBpT
33 μs to 100 s
501 × 400 pixels(one measurement diagram);max. two measurement diagramswith independent settings
1 dB, 10 dB to 200 dB in steps of 10 dB
10 % of reference level per level division,
10 divisions
3
6
max peak, min peak, auto peak, sample,
quasi-peak, average, RMS
clear/write, max hold, min hold, average
501
125 to 8001 in steps of approx. a factor of 2

Version 03.00, June 2009

Setting range of reference level
Units of level axis
Max. uncertainty of level measurement
Reference level uncertainty at 128 MHz

Frequency response referenced to
128 MHz

Uncertainty of attenuator setting

Uncertainty of reference level setting
Log/lin display nonlinearity

Bandwidth switching uncertainty

Total measurement uncertainty(95 % confidence level)

Quasi-peak indication

logarithmic level display
linear level display
logarithmic level display
linear level display–130 dBm to 30 dBm in steps of 0.1 dB
70.71 nV to 7.07 V in steps of 1 %dBm, dBmV, dBμV, dBμA, dBpW
mV, μV, mA, μA, nW, pW

level = –30 dBm, RF attenuation 10 dB, RBW 10 kHz, reference level –25 dBm
without preselection/preamplifier<0.2 dB (σ = 0.07 dB)
with preselection/preamplifier<0.3 dB (σ = 0.1 dB)
without preselection/preamplifier, AC-coupled®
R&S ESCI
9 kHz to 50 kHz< +0.5 dB/–1 dB, nominal
50 kHz to 3 GHz< 0.5 dB (σ = 0.17 dB)®
R&S ESCI7
1 MHz to 3 GHz< 0.5 dB (σ = 0.17 dB)
3 GHz to 7 GHz< 2 dB (σ = 0.7 dB)
without preselection/preamplifier, DC-coupled®
R&S ESCI
9 kHz to 3 GHz< 0.5 dB (σ = 0.17 dB)®
R&S ESCI7
9 kHz to 3 GHz< 0.5 dB (σ = 0.17 dB)
3 GHz to 7 GHz< 2 dB (σ = 0.7 dB)
with preselection/preamplifier, AC-coupled®
R&S ESCI
9 kHz to 50 kHz< +0.8 dB/–1.3 dB, nominal
50 kHz to 3 GHz< 0.8 dB (σ = 0.27 dB)®
R&S ESCI7
1 MHz to 3 GHz< 0.8 dB (σ = 0.27 dB)
3 GHz to 7 GHz< 2 dB (σ = 0.7 dB)
with preselection/preamplifier, DC-coupled®
R&S ESCI
9 kHz to 3 GHz< 0.8 dB (σ = 0.27 dB)®
R&S ESCI7
9 kHz to 3 GHz< 0.8 dB (σ = 0.27 dB)
3 GHz to 7 GHz< 2 dB (σ = 0.7 dB)< 0.2 dB (σ = 0.07 dB)
f = 128 MHz,
0 dB to 70 dB, referenced to 10 dB
RF attenuation< 0.2 dB (σ = 0.07 dB)
S/N > 16 dB
RBW ≤ 120 kHz
0 dB to –70 dB< 0.2 dB (σ = 0.07 dB)–70 dB to –90 dB< 0.5 dB (σ = 0.17 dB)
RBW > 120 kHz
0 dB to –50 dB< 0.2 dB (σ = 0.07 dB)–50 dB to –70 dB< 0.5 dB (σ = 0.17 dB)
referenced to RBW = 10 kHz
10 kHz to 120 kHz< 0.1 dB (σ = 0.03 dB)
300 kHz to 10 MHz< 0.2 dB (σ = 0.07 dB)
FFT filter, 1 Hz to 3 kHz< 0.2 dB (σ = 0.07 dB)
Signal level 0 dB to –70 dB below reference level, S/N > 20 dB, RBW ≤ 120 kHz,
DC-coupledwithout preselection/preamplifier< 3 GHz0.5 dB
3 GHz to 7 GHz
1.5 dB
with preselection/preamplifier< 3 GHz
1 dB
3 GHz to 7 GHz
1.5 dB
in line with CISPR 16-1-1®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver7

Version 03.00, June 2009

Trigger functions
Trigger
Trigger source
Trigger offset

span ≥ 10 Hz
span = 0 Hz

Max. deviation of trigger offset

Gated sweep
Gate source
Gate delay
Gate length

free run, video, external, IF level
125 ns to 100 s, resolution min. 125 ns(or 1 % of offset)±(125 ns to 100 s), resolution min. 125 ns,
dependent on sweep time±(125 ns + (0.1 % × trigger offset))video, external, IF level1 μs to 100 s
125 ns to 100 s, resolution min. 125 ns(or 1 % of gate length)± (125 ns + (0.1 % × gate length))Max. deviation of gate length

Audio demodulation
AF demodulation modes
Audio output
Marker hold time in analyzer mode

selectable

AM and FM
loudspeaker and earphone jack
100 ms to 60 s

Inputs and outputs (front panel)
RF input
Impedance
Connector
VSWR

50 Ω
N female
RF attenuation < 10 dB, DC-coupled®
®
R&S ESCI, R&S ESCI79 kHz to 1 GHz
1 GHz to 3 GHz®
R&S ESCI7
3 GHz to 7 GHz
RF attenuation ≥ 10 dB, DC-coupled®
®
R&S ESCI, R&S ESCI79 kHz to 1 GHz
1 GHz to 3 GHz®
R&S ESCI7
3 GHz to 7 GHz
RF attenuation < 10 dB, AC-coupled®
R&S ESCI
9 kHz to 100 kHz
100 kHz to 1 GHz
1 GHz to 3 GHz®
R&S ESCI7
1 MHz to 5 MHz
5 MHz to 1 GHz
1 GHz to 7 GHz
RF attenuation ≥ 10 dB, AC-coupled®
R&S ESCI
9 kHz to 100 kHz
100 kHz to 1 GHz
1 GHz to 3 GHz®
R&S ESCI7
1 MHz to 5 MHz
5 MHz to 1 GHz
1 GHz to 3 GHz
3 GHz to 7 GHz

Setting range of attenuator

Probe power supply
Supply voltages

8< 2.0, typ. 1.5< 3.0, typ. 2.5< 3.0, typ. 2.5

< 1.2
< 1.5
< 2.0

2.5
2.0
3.0
2.5
2.0
3.0

typ. 2.5
< 1.2
< 1.5
typ. 2.5
< 1.2
< 1.5
< 2.0
0 dB to 70 dB in steps of 5 dB

+15 V DC, –12.6 V DC and ground,max. 150 mA, nominal®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver

Version 03.00, June 2009

Power supply for antennas, etc.
Supply voltages±10 V DC and ground,max. 100 mA, nominal

USB interface

2 ports, type A plug, version 2.0

AF output
Connector
Impedance
Open-circuit voltage

3.5 mm jack
10 Ω
adjustable up to 1.5 V

Inputs and outputs (rear panel)
IF 20.4 MHz
Connector
Impedance
Level

BNC female
50 Ω
mixer level > –60 dBm
RBW ≤ 100 kHz or FFT
RBW > 100 kHz–10 dBm at reference level
0 dBm at reference level

Reference frequency output
Connector
Impedance
Output frequency
Level

BNC female
50 Ω
10 MHz0 dBm, nominal

Reference frequency input
Connector
Input frequency
Required level

BNC female
10 MHz0 dBm from 50 ΩPower supply for noise source
Connector
Output voltage

switchable

External trigger/gate input
Connector
Impedance
Trigger voltage
IEC/IEEE bus remote control

BNC female
28 V, nominal

BNC female> 10 kΩ1.4 V (TTL)interface in line with IEC 625-2(IEEE 488.2)
24-pin Amphenol female
SCPI 1997.0
SH1, AH1, T6, SR1, RL1, PP1, DC1, DT1,
C0

Connector
Command set
Interface functions

Serial interface

RS-232-C (COM), 9-pin D-Sub

Printer interface

parallel (Centronics compatible)

USB interface

upper connector
lower connector

type A plug, version 1.1
type A plug, version 2.0

External monitor (VGA)
Connector

VGA-compatible, 15-pin D-Sub

User interface

25-pin D-Sub®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver

9Version 03.00, June 2009

General data
21 cm TFT color display
640 × 480 pixel (VGA)–5< 2 × 10

Display
Resolution
Pixel error rate

Mass memory
Data storage®

R&S ESCI only

Temperature ranges
Operating temperature range®

with R&S ESCI-B20 option
Permissible temperature range®

with R&S ESCI-B20 option
Storage temperature range
Climatic loading

Mechanical resistance
Sinusoidal vibration

Random vibration®

with R&S ESCI-B20 option
Shock

Recommended calibration interval

operation with external reference
operation with internal referencePower supply
AC supply+5 °C to + 40 °C
0 °C to +50 °C+5 °C to + 45 °C
0 °C to +55 °C–40 °C to +70 °C+40 °C at 95 % relative humidity(EN 60068-2-30)

0.5 g from 5 Hz to 150 Hz,
max. 2 g at 55 Hz,
in line with EN 60068-2-6, EN 61010-1,
MIL-T-28800D, class 5
10 Hz to 100 Hz, acceleration 1 g (rms)
10 Hz to 300 Hz, acceleration 1.9 g (rms)
40 g shock spectrum,in line with MIL-STD-810C and
MIL-T-28800D, classes 3 and 5
2 years
1 year100 V to 240 V AC, 50 Hz to 400 Hz,
3.1 A to 1.3 A,
class of protection I in line with VDE 411
typ. 70 VA
in line with EN 61010-1, UL 3111-1,
CSA C22.2 No. 1010-1, IEC 1010-1
EMC Directive 2004/108/EC
including:
EN 61326 class B (emission),
CISPR 11/EN 55011 group 1 class B(emission)
EN 61326 table A.1 (immunity,
industrial)
VDE, GS, CSA, CSA-NRTL/C

Power consumption
SafetyEMC

Test marks

Dimensions and weight
Dimensions

W×H×D
®

Weight without options

10

1.44 Mbyte 3½'' disk drive, hard disk
> 500 instrument setups and traces

R&S ESCI®
R&S ESCI7®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver

412 mm × 197 mm × 417 mm(16.22 in × 7.76 in × 16.42 in)
10.5 kg (23.15 lb)
12.4 kg (27.34 lb)

Version 03.00, June 2009

Ordering information
DesignationType®
EMI Test Receiver 9 kHz to 3 GHz
R&S ESCI®
EMI Test Receiver 9 kHz to 7 GHz
R&S ESCI7
Accessories suppliedPower cable, operating manual, service manual

Order No.
1166.5950.03
1166.5950.07

Options
Designation
Rugged Case, with carrying handle
OCXO Reference Frequency
TV Trigger/RF Power Trigger
Internal Tracking Generator, I/Q Modulator
External Generator Control
LAN Interface 100BaseT
Expanded Environmental Specifications
DC Power Supply
Battery Pack
Spare Battery PackType®
R&S FSP-B1®
R&S FSP-B4®
R&S FSP-B6®
R&S FSP-B9®
R&S FSP-B10®
R&S FSP-B16®
R&S ESCI-B20®
R&S FSP-B30®
R&S FSP-B31®
R&S FSP-B32

Order No.
1129.7998.02
1129.6740.02
1129.8594.02
1129.6991.02
1129.7246.03
1129.8042.03
1155.1606.14
1155.1158.02
1155.1258.02
1155.1506.02

Type

Order No.

Service Options
Designation®
R&S ESCI
One-Year Repair Service
following the warranty period
Two-Year Repair Service
following the warranty period
Four-Year Repair Service
following the warranty period
Two-Year Calibration Service
Three-Year Calibration Service
Five-Year Calibration Service®
R&S ESCI7
One-Year Repair Service
following the warranty period
Two-Year Repair Service
following the warranty period
Four-Year Repair Service
following the warranty period
Two-Year Calibration Service
Three-Year Calibration Service
Five-Year Calibration Service®

1166.5950.S16®

1166.5950.S12®

1166.5950.S14®

1166.5950.S15
1166.5950.S11
1166.5950.S13®

1166.5950.S26®

1166.5950.S22®

1166.5950.S24®

1166.5950.S25
1166.5950.S21
1166.5950.S23

R&S RO2ESCIR&S RO3ESCIR&S RO5ESCIR&S CO2ESCI®
R&S CO3ESCI®
R&S CO5ESCIR&S RO2ESCI7
R&S RO3ESCI7
R&S RO5ESCI7
R&S CO2ESCI7®
R&S CO3ESCI7®
R&S CO5ESCI7

For product brochure, see:••PD 0758.1558.12 (ESCI)PD 5214.2762.12 (ESCI7)

and www.rohde-schwarz.com®

Rohde & Schwarz R&S ESCI/ESCI7 EMI Test Receiver

11Service you can rely on
J Worldwide
J Local and personalized
J Customized and flexible
J Uncompromising quality
J Long-term dependability

About Rohde & Schwarz
Rohde & Schwarz is an independent group of companies
specializing in electronics. It is a leading supplier of solutions in the fields of test and measurement, broadcasting,radiomonitoring and radiolocation, as well as secure communications. Established 75 years ago, Rohde & Schwarz
has a global presence and a dedicated service network inover 70 countries. Company headquarters are in Munich,Germany.
Regional contact
Europe, Africa, Middle East
+49 1805 12 42 42* or +49 89 4129 137 74
customersupport@rohde-schwarz.com
North America
1 888 TEST RSA (1 888 837 87 72)
customer.support@rsa.rohde-schwarz.com
Latin America+1 410 910 79 88
customersupport.la@rohde-schwarz.com
Asia/Pacific+65 65 13 04 88
customersupport.asia@rohde-schwarz.com

Certified Quality System

ISO 9001

Certified Environmental System

ISO 14001

Rohde & Schwarz GmbH & Co. KG
Mühldorfstraße 15 | 81671 München
Phone +49 89 41 290 | Fax +49 89 41 29 121 64
www.rohde-schwarz.com
R&S® is a registered trademark of Rohde & Schwarz GmbH & Co. KG
Trade names are trademarks of the owners | Printed in Germany (sv)
PD 0758.1558.22 | Version 03.00 | June 2009 | R&S®ESCI/ESCI7
Subject to change*0.14 €/min within German wireline network; rates may vary in other­networks (wireline and mobile) and countries.