Precision Zero Air
Zero Air Generator for GC
Part Number : See Reverse

Service Kit : 08-3611

Your local gas generation partner

Description
The Precision Zero Air generators are designed specifically to supply
clean, dry, hydrocarbon-free air to be used as detector gas for GC.
Five models are available, delivering 1.5, 3.5, 7, 18 and 30 liters per
minute flow rate, dependent upon the specific requirements.
As with all Precision Series generators, Zero Air benefits from acompact and modular, stackable design, minimizing the total footprint
required for GC gas supply, and providing flexibility to add or remove
modules as your laboratory requirements evolve over time. There arethree module heights, the 1.5L and 3.5L come in the smaller unit, the 7L
and 18L in the medium with the 30L generator housed in a larger unit.

Applications)

Detector Gas• Flame Support / Air

Key Features

540 mm

380 mm

156 mmZero Air 1.5L/3L unit size pictured in a typical Precision stack.

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Zero Air Datasheet Rev 13 RSID 386 EN —10/09/2024)

Integrates seamlessly with other Precision units)

Hydrocarbon content <0.05ppm for market-leadingpurity)

Minimum lifetime maintenance requirements, no
expensive catalyst chamber replacements)

Avoid risk of contaminants entering the system (whenswitching out empty cylinders))No risk of running out of gas in middle of analysis)

Compatible with Precision Air Compressor)

12 month comprehensive on-site warranty)

Status indicative lighting feature (amber warm-up,green ready)Technical Specifications
Max Flow Rate

Zero Air 1.5L

Zero Air 3.5L

Zero Air 7L

Zero Air 18L

Zero Air 30L

1.5L/min3.5L/min

7L/min

18L/min

30L/minMax Pressure

80 psi / 5.5 bar

100 psi / 6.9 bar

Hydrocarbon Concentration(as Methane)<0.05ppm

Gas Outlets

1 x 1/4" BSPP
90-145 psi
6.2-10 bar

Min/Max Air Inlet Pressure
Min Air Inlet Flow

1.5L/min110-125 psi
7.5-8.6 bar
3.5L/min

7L/minMin Inlet Air Quality

110-145 psi
7.6-10 bar
18L/min

30L/min

ISO8573 - 1:2010 Class 1.4.1

Phthalates

None

Dew Point-30°C ( -22 °F )Suspended Liquids

NoneStart-Up Time For Purity

60 minutes

Operating Temperature

5°C - 35°C / 41°F - 95°F

Electrical Requirements

230VAC 50/60 Hz 2A
110VAC 50/60Hz 1A

230VAC 50/60Hz 5.4A
110VAC 50/60 Hz 3.3A

230VAC 50/60 Hz 11A
110VAC 50/60Hz 6.7A

Power Consumption

220 Watts @ 110VAC
or
230 Watts@ 230VAC

594 Watts @ 110VAC
or
759 Watts @ 230VAC

1210 Watts @ 110VAC
or
1541 Watts @ 230VAC

Heat Output

up to 785 BTU/Hr

up to 2590 BTU/Hr

up to 5255 BTU/Hr

Generator Dimensions (WxDxH)

38.0 x 54.0 x 15.6 cm
14.9 x 21.2 x 6.14"

38.0 x 54.0 x 25.6 cm
14.9 x 21.2 x 10”

38.0 x 54.0 x 40.5 cm
14.9 x 21.2 x 15.9"

16 Kg / 35 lbs

25kg / 55.1 lbs

41 Kg / 90.3 lbs

Generator Weight

Ordering Information
Part Number

Zero Air 1.5L

Zero Air 3.5L

Zero Air 7L

Zero Air 18L

Zero Air 30L

110 V

230 V

110 V

230 V

110 V

230 V

110 V

230 V

110 V

230 V

60-150160-1500

60-350160-3500

60-0008

60-0007

60-001960-001860-003160-0030

Annual Service Kit
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan

Precision Air Compressor

110v

230v

Annual Service Kit

Part Number

65-1555

65-055508-8372

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a comprehensive on-site 12 month warranty. Beyond this period however you can ensure that you’reinvestment continues to be [Protected] by our extensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UKTel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management Systemconforms to: ISO:9001:2008*For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: discover@peakscientific.com
Copyright © Peak Scientific Instruments Ltd 2024 — Precision Zero Air Datasheet Rev 13 RSID 386 EN —10/09/2024* CSA certification applies to 1.5L, 3.5L & 30L models only.