ORBIT 60 SERIES

Dynamic Input Modules
Datasheet

Bently Nevada Machinery Condition Monitoring

137M0698 Rev. F

Description
The primary purpose of the Dynamic Input module is todigitize the sensor signal at a rate that completely
encompasses the signal content and provides transducer
power for various sensors. The Orbit 60 Series Dynamic Input
modules are a set of 4-channel input modules available inboth negative and positive dynamic input options. The inputs
are also used for speed or Keyphasor signals.
The PAV, PAS, PAA, PAD and PVT modules can be
configured with up to TWO SPEED CHANNELS with amaximum speed of 12,000 rpm and maximum
speed impulse rate of 12,000 cpm (200 Hz). For morethan two speed channels on a single dynamic input
card, speeds greater than 12,000 rpm or speed
impulse frequencies greater than 12,000 cpm (200Hz) a KPH Module is needed.
The Orbit 60 dynamic input modules are designed for use on
a broad range of machine trains or individual casings where
the sensor point count fits the monitor’s channel count andwhere advanced signal processing is desired. The modules
are optimized for intensive signal processing required oncomplex machinery such as gearboxes, planetary gearboxes,
reciprocating compressors, and roller element bearing (REB)
machines as well as offering advanced measurement
capabilities on conventional monitoring methods such asradial vibration, thrust position, piston rod monitoring, andcasing absolute vibration.

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Negative Transducer Input Modules
The following cards work with negative-voltage
external sensors offering four variants:
l

l

l

l

l

PAV Negative Dynamic Sampler (Prox,
Accel, Velom)
PAS Negative Dynamic Sampler (Prox,
Accel, Seismic)
PAA Negative Dynamic Sampler (Prox,
Accel, Aero)
PAD Negative Dynamic Sampler (Prox,
Accel, DC LVDT)KPH High Speed Keyphasor (Prox, Accel,
Magnetic Pickup)

Positive Transducer Input Module
The Positive Voltage Transducer (PVT) input
module interfaces with industry-standard
third-party IEPE sensors, as well as sensors thatuse a 3-wire (power, common, signal) or acustom 2-wire (A/+ and B/-) positive-voltage
interface.
The PVT is the preferred module to use for IEPE
sensors, including the Bently Nevada Velomitor(3005xx) and IEPE accelerometers. Using the
PVT modules for these sensors improves noise
performance of the sensor.
l

PVT Positive Dynamic Sampler (Prox,
Accel, Velom)

The PVT module is recommended for new
Velomitor installations only. Projects using the190501 Velomitor CT or retrofits that reuse otherexisting Velomitor sensors should use the PAV
module unless the user can verify the sensor
power limits are appropriate for existing
Velomitors.

Connectors
The Dynamic Input module uses an ix Industrial
connection to provide access to four bufferedtransducer output (BTO) connectors for each of
the dynamic channels, with short circuit
protection. The ix Industrial connection isavailable on the public and utility side of themodule.

ORBIT 60 SERIES Dynamic Input Modules Datasheet

2/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Dynamic Input Modules

Dynamic Input Modules

PAS

Prox/Accel (3-wire)
0-40 kHz 1% of Full Scale

Dynamic Input Modules
PAV(-) (Prox, Accel, Velom)

Seismic (2-wire)PAS(-) (Prox, Accel, Seismic)

PAA(-) (Prox, Accel, Aero)

5 Hz - 20 kHz 1% of Full
Scale

PAD

(-) (Prox, Accel, DC LVDT)

PVT(+) (Prox, Accel, Velom)

20-40 kHz 2% of Full Scale
PAA

Prox/Accel (3-wire)
0-40 kHz 1% of Full Scale

Speed and Keyphasor

Aero (4-wire)

Speed Range

1 - 12,000 ppm (pulses per
minute)5 Hz - 20 kHz 1% of Full
Scale

Minimum Pulse
Width

Keyphasor Pulse Width
must be greater than orequal to 10 microseconds.

20-40 kHz 2% of Full Scale
PAD

0-40 kHz 1% of Full Scale

Power Consumption
Maximum

11 W

Typical(All Modules)

7.5 W

DC LVDT (4-wire)
5 Hz - 20 kHz 1% of FullScale
20-40 kHz 2% of Full Scale
PVT

Accuracy and Frequency Response
PAV

Prox/Accel (3-wire)

0-40 kHz 1% of Full Scale

Prox/Accel (3-wire)

Velom (2-wire)

0-40 kHz 1% of Full Scale

5 Hz - 20 kHz 1% of Full
Scale

Velom (2-wire)
5 Hz - 20 kHz 1% of Full
Scale

Recommended top scale= 1 in/s to meet 1%
accuracy

Recommended top scale= 1 in/s to meet 1%
accuracy

20-40 kHz 2% of Full Scale
Dynamic Inputs

20-40 kHz 2% of Full Scale

ORBIT 60 SERIES Dynamic Input Modules Datasheet

Prox/Accel (3-wire)

3/19

Analog Input

See Input Module Sensors
and Channels on page 8.

Channels
Supported

4 Dynamic Inputs

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Dynamic Input Modules
Sampling Rate

Dynamic Input Modules

102.4 kHzPAA

Min. -22, Max. 0

Input Interface Impedance (Typical)
PAV

PAS

PAA

PAD

Prox/Accel (3-wire)

Aero (4-wire)

10 kΩ

Min. -22, Max. 0
PAD

Prox/Accel (3-wire)

Min. -22, Max. 0

Seismic (2-wire)

DC LVDT (4-wire)

10 kΩ

Min. -10, Max. 10
PVTProx/Accel (3-wire)

Min. 0, Max. 24

Aero (4-wire)

Velom (2-wire)

100 kΩ

Min. 2, Max. 24

Prox/Accel (3-wire)

Outputs

10 kΩ

1 MΩ

Analog Buffered
Short circuit protectedTransducer (BTO) output signal available
through BTO connector on
public and utility side.

Prox/Accel (3-wire)

BTO AccuracyAC> 0 to < 10 kHz, ±1% of input
signalInput Interface Signal Range [V]

10 kHz to < 20 kHz, ±2% of
input signal

Prox/Accel (3-wire)

20 kHz to < 30 kHz, ±4% of
input signal

Min. -22, Max. 0
Velom (2-wire)PAS

Prox/Accel (3-wire)

10 kΩ

10 kΩ

PAV

Prox/Accel (3-wire)

10 kΩ

DC LVDT (4-wire)

PVT

Prox/Accel (3-wire)

Min. -24, Max. -2

30 kHz to ≤ 40 kHz, ±6% of
input signal

Prox/Accel (3-wire)

DC

Min. -22, Max. 0±100 mV over voltage
range of Input Module

Seismic (2-wire)
BTO Output
Impedance

Min. -14, Max. 0

ORBIT 60 SERIES Dynamic Input Modules Datasheet

4/19

500 Ω

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Dynamic Input Modules

Dynamic Input Modules

BTO Connector

This is a true analog signal from the
input, not digital to analog
reconstitution of the input
signal.Some Transducers have an
offset BTO bias.

Module OK LED

Indicates when themodule is functioning
properly

SystemCommunication
LED

indicates when themodule is communicating
to the rest of the system

PhysicalRequired Rack
SpaceTransducer Power
PAV

Environmental Limits

Prox/Accel (3-wire)

ChassisOperatingTemperature
Range-24 VDC, Max. 40 mA
Velom (2-wire)
3.3 mA (Constant current)PAS(indoor use
only)

Prox/Accel (3-wire)-24 VDC, Max. 40 mA

PAA

1 SlotProx/Accel (3-wire)-24 VDC, Max. 40 mA

3U Chassis:-30°C to +70°C(-22°F to 158°F)
6U Chassis:-30°C to +65°C(-22°F to 149°F)
Temperatures over
50°C (122°F) require
forced air convection
with a minimum
airspeed of 0.5 m/s.

Aero (4-wire)
ModuleTemperature
Rating
Certification-24 VDC, Max. 40 mA
PAD

Prox/Accel (3-wire)-24 VDC, Max. 40 mA
DC LVDT (4-wire)-10 to 10 VDC, max. 40 mA

PVTProx/Accel (3-wire)
24 VDC, Max. 33 mA
Velom (2-wire)LEDs
1 per input channel
indicates when theconnected sensor is in an
OK condition

ORBIT 60 SERIES Dynamic Input Modules Datasheet

When using a
Bridge module,
temperatures over
58°C (136°F) require
forced air
convection with aminimum airspeedof 0.5 m/s.
You must still meet
the Chassis
OperatingTemperature Range
defined above.

9.5 mA (Typical)

Channel Status
LED(Rear Utility side
only)

-30°C to +70°C(-22°F to 158°F)

5/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Environmental Limits
Storage
Temperature
Range-40°C to +85°C(-40°F to 185°F)

Relative
Humidity

0% to 95% rH non-condensing
operating and storage

Vibration

Without Isolators:
0 g to 0.35 g @ 57-500 Hz

CAUTIONLOCATION TEMPERATURE
AND HUMIDITY
While the system has beentested and capable of achieving
the design life when operating inenvironments up to 70°C,
whenever operating any
electronics system in elevatedhumidity or temperatures
exceeding 40°C, adding
environmental controls
maximizes the operational life of
the system.

With Isolators:
0 g to 5 g @ 57-500 Hz
Shock

2” Incline Drop

Altitude< 2000 m (6,562 ft)
Higher altitudes are
possible but are site
specific
applications.
Contact Bently
Nevadasupport if you
require higher
altitudes.

Pollution
DegreePollution Degree 2

Installation
Category

Category II

Verify that temperature ratings on thewiring cables match the operating
temperature range.

ORBIT 60 SERIES Dynamic Input Modules Datasheet

6/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Compliance and Certifications*Maritime
ABS Rules for Condition of Classification,
Part 1

FCC
This device complies with part 15 of the
FCC Rules. Operation is subject to thefollowing two conditions:
l

l

l
l

Steel Vessels Rules
Offshore Units and Structures*Recorder Output module, Bridge
module, and 6U systems
approvals pending

This device may not cause harmfulinterference.
This device must accept any
interference received, includinginterference that may cause
undesired operation.

Functional Safety
SIL 2
See the SIL User Guide (134M0398) for
details regarding SIL implementation.

EMC
European Community Directive:

Hazardous Area Approvals

EMC Directive 2014/30/EU

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

Standards:
EN 61000-6-2; Immunity for
Industrial Environments
EN 61000-6-4; Emissions for
Industrial Environments

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

Electrical Safety
European Community Directive:
LV Directive 2014/35/EU

cNRTLusStandards:

Class I, Zone 2: AEx/Ex ec nC IIC T4 Gc;
Class I, Zone 2: AEx/Ex nA nC IIC T4 Gc;
Class I, Division 2, Groups A, B, C, D T4;
Class I, Division 2, Groups A, B, C, D T4
(N.I.);EN 61010-1;
EN 61010-2-201;

RoHS
European Community Directive:

T4 @ Ta= -30˚C to +70˚C (-22˚F to +158˚F)

RoHS Directive 2011/65/EU

ATEX/IECEx

Cyber Security

II 3 G
Ex ec nC IIC T4 Gc
Ex nA nC IIC T4 Gc

Designed to meet IEC 62443-4-2

T4 @ Ta= -30˚C to +70˚C (-22˚F to +158˚F)

ORBIT 60 SERIES Dynamic Input Modules Datasheet

7/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Input Module Sensors and Channels
Sensor Type
Supported

Channel TypeDynamic Input Module Type
(4 channels)

Static Input
Module Type
(6 channels)

PAV

PAS

PAA

PAD

PVT

KPH

X

X

X

X

X

X

Proximitor(3-wire)Differential
Expansion, Radial
Vibration, Speed,
Thrust, Recip
Piston Rod

Magnetic Pickups

Speed

Accelerometer(3-wire)Acceleration1,
Recip Impulse
Acceleration

X

X

X

X

X2

X

Charge Amplifier(3-wire)Acceleration1

X

X

X

X2

X2

X

Interface Modules (4wire)

Acceleration1

X

High-Temp Accel(4-wire)Acceleration1

X

High-Temp Accel(3-wire)Acceleration1

X

X

X2

X

Negative Biased
Constant Current (2wire)

Acceleration1

X

IEPE Positive ConstantCurrent (2- wire)

Acceleration1,
Recip Impulse
Acceleration

High-Temp Velocity

Velocity 1

X

Negative Biased
Constant Current (2wire)

Velocity 1

X

Velomitor®
(2-wire)Velocity 1

X2, 3

Velomitor CT

Velocity 1

X2, 3

X

Velocity 1

Amplifier/Interface
Modules

Dynamic Pressure

X

X

Seismoprobe (2-wire) Velocity 1
IEPE Positive ConstantCurrent (2- wire)

X

X

X

X2

X2, 3

X
X3

X
X

8/19

AC
LVDT

Temp

PVD

ORBIT 60 SERIES Dynamic Input Modules
Datasheet
Sensor Type
Supported

137M0698 Rev. F

Channel TypeDynamic Input Module Type
(4 channels)
PAV

PAS

PAA

PAD

PVT

KPH

Static Input
Module Type
(6 channels)

AC
LVDT

Temp

PVD

Pressure Transducers

Dynamic PressureX

DC LVDT

Valve Position &Case Expansion

AC LVDT

Valve Position &Case Expansion

3-wire RTD

Temperature

X

TC - Type J, K, E, T

TemperatureX

4-20 mA Transmitter,±10 V Sensor

Process VariableX

Dry or Wet Contact,
TTL Logic

Discrete Channel

X

X

X

1 Designates the ability to integrate these measurements to provide additional measurement types.
2 These sensors can be configured using a Custom transducer configuration.
3 PVT modules are recommended for new sensor installations only. Projects using the Velomitor CT or retrofits that reuse

existing sensors should use PAV or verify sensor power limits.

The PVT is only for positively biased sensors.
The Velomitors® and IEPE sensors can be configured on the PAV as a custom transducer.

9/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Ordering
OptionOrdering Information

B – SIL Level

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

AAA – Hazardous Area Certifications

CSA/NRTL/C (Class I, Div 2)

02

Multi (CSA, ATEX, IECEx)

XXXCountry Specific Approvals

No SIL2

SIL 2

PAA (Prox/Accel/Aero) Module
Ordering
OptionDescription

60R/INP02-AAA-B
AAA – Hazardous Area Certifications
00

No Hazardous Area

01

CSA/NRTL/C (Class I, Div 2)

02

Multi (CSA, ATEX, IECEx)

XXXCountry Specific Approvals

ORBIT 60 SERIES Dynamic Input Modules Datasheet

Description

00

No Hazardous Area

01

CSA/NRTL/C (Class I, Div 2)

02

Multi (CSA, ATEX, IECEx)

XXXCountry Specific Approvals

B – SIL Level

B – SIL Level
0

SIL 2

AAA – Hazardous Area Certifications

60R/INP01-AAA-B

01

2

60R/INP03-AAA-B

Description

No Hazardous Area

No SIL

Ordering
Option

PAV (Prox/Accel/Vel) Module000

PAS (Prox/Accel/Seismic) Module

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

Ordering
OptionDescription

10/19

0

No SIL2

SIL 2

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

PAD (Prox/Accel/DCLVDT) Module
Ordering
OptionExternal Barriers

Description

Part
NumberDescription

60R/INP04-AAA-B

1755023-pin Transducer Barrier

AAA – Hazardous Area Certifications

1772412-pin Velomitor Barrier

00

No Hazardous Area

Thermocouple Barrier

01

CSA/NRTL/C (Class I, Div 2)175990 or
170M3559

02

Multi (CSA, ATEX, IECEx)

170M3559

RTD Barrier

XXX

Country Specific Approvals

External Galvanic Isolators
Part
NumberB – SIL Level
0

No SIL2

SIL 2

PVT (Prox/Accel/Velom)
Ordering
OptionDescription

60R/INP05-AAA-B
AAA – Hazardous Area Certifications
00

No Hazardous Area

01

CSA/NRTL/C (Class I, Div 2)

02

Multi (CSA, ATEX, IECEx)

XXXCountry Specific Approvals

B – SIL Level
0

No SIL2

SIL 2

Accessories
Part
NumberDescription

60X/BTC01

Buffered Transducer Breakout
Kit

ORBIT 60 SERIES Dynamic Input Modules Datasheet

11/19

Description

103M7134

3-pin Transducer Isolator

103M71342-pin Transducer Isolator

154M1361

Thermocouple Isolator

103M7138

RTD Isolator

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Dynamic Input Module Layout
PAV

PAA

Public Side - Utility Side

Public Side - Utility Side

12/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

PAS

PAD

Public Side - Utility Side

Public Side - Utility Side

13/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

PVT

Public Side - Utility Side

14/19

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Negative Dynamic Sensor
Interfaces

Radial Vibration, Thrust and Speed
Measurements

These modules accept one to four sensor
inputs.
Radial Vibration, Thrust and Speed
Measurements
3300 5 mm, 5M

Proximity
Transducer
System

3300 5 mm, 9M

Proximity
Transducer
System

3300 8 mm, 5M

Proximity
Transducer
System

3300 8 mm, 9M

Proximity
Transducer
System

3300 HTPSHigh Temperature
Proximity System

3300 RAM

Radiation
Resistant Proximity
Transducer

3300 (0.3 in - 15 ft)

Radiation
Resistant Proximity
Transducer

3300 RAD (0.3 in - 110 ft)

Radiation
Resistant Proximity
Transducer

3300 (0.42 in - 15 ft)

Radiation
Resistant Proximity
Transducer

7200 5 mm

Proximity System

7200 8 mm

Proximity System

7200 11 mm

Proximity System

ORBIT 60 SERIES Dynamic Input Modules Datasheet

Proximity System

3300XL NSV

Proximity System

3300XL 5 mm, 5M

Proximity System

3300XL 5 mm, 9M

Proximity System

3300XL 8 mm

Proximity System

3300XL 8 mm, 9M

Proximity System

3300XL 11 mm

Proximity System

Magnetic Pickup (MPU)

General Magnetic
Pickup Speed
Sensor (Supported
only on Keyphasor
Input Module)

Custom Proximitor

Allows User
Definition

Acceleration Measurements

Radiation
Resistant Proximity
Transducer

3300 (0.3 in - 40 ft)

7200 14 mm

15/19

200350Accelerometer

200355

Accelerometer

23733-03

Accel I/F Module

24145-02

High-Freq Accel I/F Module

330400100 mV/g Accelerometer

33042525 mV/g Accelerometer

330450High Temp Accelerometer

350501

Acceleration Charge Amplifier

350900HTVAS High Temp Velocity and
Accel Sensor

3700300

Accelerometer

86517

Accel Interface Module

CustomAllows User Definition

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Velocity Measurements
9200

Seismoprobe

74712

High Temp Seismoprobe

47633

Seismoprobe

86205

Velocity Transducer (Mag coil
design)190501

Velomitor CT (PAV Only)

330500

Velomitor

330505

Low Freq Velocity Sensor

330525

Velomitor

330530

Radiation Resistant Velomitor

330750High Temp Velocity Sensor

330752High Temp Velocity Sensor

350900 HTVASHigh Temp Velocity & Accel
Sensor86517

Accelerometer Interface
ModuleCustomSeismoprobe

Allows User Definition

CustomAllows User Definition

Case Expansion and Valve Position
Measurements
3300XL RPT

RotaryPositionTransducer
System(ValvePosition Only)

24765-01 DC LVDTDC Linear
Variable
Differential
Transformer(PAD Only)

24765-02 DC LVDTDC Linear
Variable
Differential
Transformer(PAD Only)

24765-03 DC LVDTDC Linear
Variable
Differential
Transformer(PAD Only)

135613-01 High Temperature
DC LVDT

DC Linear
Variable
Differential
Transformer(PAD Only)

Dynamic Pressure Measurements

86517 withMOD
159840

Dynamic Pressure Interface
Module165855

Cylinder Pressure Transducer

350300Pressure Dynamic Sensor

350500

Dynamic Pressure Charge
AmplifierCustomPressure
SensorAllows User Definition

ORBIT 60 SERIES Dynamic Input Modules Datasheet

135613-11 High Temperature DC DC Linear
LVDT
Variable
Differential
Transformer(PAD Only)
135613-02 High Temperature
DC LVDT

16/19

DC Linear
Variable
Differential
Transformer(PAD Only)ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Case Expansion and Valve Position
Measurements
135613-12 High Temperature
DC LVDT

DC Linear
Variable
Differential
Transformer(PAD Only)

135613-03 High Temperature
DC LVDT

DC Linear
Variable
Differential
Transformer(PAD Only)

135613-13 High Temperature
DC LVDT

DC Linear
Variable
Differential
Transformer(PAD Only)

18639-01 +/-0.5in AC LVDTAC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

18639-02 +/-1 in AC LVDT18639-03 +/-0.531 in AC LVDT18639-04 +/-6in AC LVDT

Case Expansion and Valve Position
Measurements

AC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)
AC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

18639-05 +/-2in AC LVDTAC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

18639-06 +/-3in AC LVDTAC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

18639-07 +/-5in AC LVDTAC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

18639-08 +/-10in AC LVDTAC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

18639-09 +/-4in AC LVDTAC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

Custom Proximitor

Allows User
Definition(Case
Expansion
Only)

AC Linear
Variable
Differential
Transformer(AC LVDT
Module Only)

Differential Expansion
3300 5mm,
5M

ORBIT 60 SERIES Dynamic Input Modules Datasheet

17/19

Proximity Transducer System

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Differential Expansion
3300 5mm,
9M

Proximity Transducer System

3300 8mm,
5M

Proximity Transducer System

3300 8mm,
9M

Proximity Transducer System

3300 HTPSHigh Temperature Proximity
System

7200 5mm

Proximity Transducer System

7200 8mm

Proximity Transducer System

7200 11mm

Proximity Transducer System

7200 14mm

Proximity Transducer System

3300XL5mm, 5M

Proximity Transducer System

3300XL5mm, 9M

Proximity Transducer System

3300XL
8mm, 5M

Proximity Transducer System

3300XL
8mm, 9M

Proximity Transducer System

3300XL
11mm

Proximity Transducer System

3300XL
25mm

Proximity Transducer System

3300XL
50mm

Proximity Transducer System

CustomProximitor

Allows User Definition

Recip Impulse Acceleration

Proximity Probe

3300 XL
11 mm

Horizontal and Vertical Proximity
Probe

Custom3-Wire Transducers

ORBIT 60 SERIES Dynamic Input Modules Datasheet

Accelerometer

330425

Accelerometer

CustomTransducers

Custom Transducers
Custom transducers are software configurable
within the following ranges:
Custom Transducers
Scale factor

1mv/Eng Unit to2000 mv/Eng
Unit

Input voltage range

PVT+0 V to +23 V
All other modules+0 V to -23 V

OK checking voltage range PVT+0 V to +23
All other modules+0 V to -23 V

Recip Piston Rod
3300 XL
8 mm

33040018/19

Engineering units

Selection fromstandard units
table or custom
unit entry

350900

HTVAS

47633

Velocity
Seismoprobe

86205

Velocity
Transducer

350500Pressure Mod

86517

Custom Input

ORBIT 60 SERIES Dynamic Input Modules
Datasheet

137M0698 Rev. F

Copyright 2024 Baker Hughes Company. All rights reserved.

Bently Nevada and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes business, in the UnitedStates and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company. All other product andcompany names are trademarks of their respective holders. Use of the trademarks does not imply any affiliation
with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

19/19