Precision Hydrogen
Hydrogen Generator for GC
Part Number : See Reverse

Service Kit : See Reverse

Your local gas generation partner

Description
The Precision Hydrogen generators are designed to provide thegas needed for detectors requiring Hydrogen fuel gas, such as FID.
One generator is capable of supplying multiple detectors, and areavailable in various flow rates to ideally suit individual customers’
needs.
These generators utilise a Proton Exchange Membrane to create
the Hydrogen gas from deionized water, and a desiccant filtration
stage is used to dry the gas.
Precision Hydrogen gas generators come with various safety
features as standard, giving you complete peace of mind in thelaboratory and a far safer, dependable and more convenient
alternative to cylinder gas.

Applications)

Detector Gas• Fuel Gas

Key Features

540 mm)Suitable for fuel gas at standard detection limits)

99.9995% Purity)

Internal leak detection with automatic shutdownfeatures)

Automatic loading pump as standard)

Simple maintenance, limited to replacing de-ionizer
cartridge and silica gel)Short and easy start-up and shutdown procedures)Compact, space-saving modular design)

Creates hydrogen on demand, minimal storage ofhydrogen in the system)

Peak offers a 3 year cell warranty with this generator
as standard.

380 mm

406 mmPrecision Hydrogen shown in a typical Precision stack.

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Hydrogen Standard Data Sheet RSID 382 Rev 13 EN —10/09/2024

Technical Specifications
Max Flow Rate*

Hydrogen, 100cc

Hydrogen, 200cc

Hydrogen, 300cc

Hydrogen, 450cc

100cc/min

200cc/min

300cc/min

450cc/min

Max Pressure

100 psi / 6.9 bar

Purity99.9995%Gas Outlets

1 x 1/8” Swagelok compression fitting

Water Purity Requirement<1.0μ Siemens/cm OR >1 Mohm-cm

Water Consumption

up to 0.12L/ day

up to 0.24L/day

up to 0.36L/day

Operating Temperature

10°C - 35°C / 50°F - 95°F

Electrical Requirements

100-230V 50/60HZ 6A(100-450cc) 2.3-4.8A(1200cc)

Power Consumption

309-422 Watts

367-422 Watts

Heat Output

405-422 Watts

up to 0.53L/day

422 Watts

up to 1000 BTU/hr

Generator Dimensions (WxDxH)

38.0x 54.0 x 40.6cm / 14.9 x 21.2 x 15.9”

Generator Weight

31 Kg (68.3 lbs)

Noise Level

Silent in operation* All flow rates are stated in sccm at 273.15K and 1.01bar

Ordering Information
Part Number

63-0100

63-0200

63-0300

63-0450

Annual Service
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan

Accessories

Water Bottle 4L

Water Bottle 8L

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com
Copyright © Peak Scientific Instruments Ltd 2024 — Precision Hydrogen Standard Data Sheet RSID 382 Rev 13 EN —10/09/2024