PORTABLE APPLIANCE TESTERKEW 6205

Protect life and property fromfaulty appliancesUSB Battery operated
 PASS/FAIL result
 Colour status back light 10mA & 30mA RCD test

(Isolation transformer built in) Memory function up to 999 data Printer output
 Complies to AS/NZS 3760

Print out measurement results

● KEW 6205 Function● KEW 6205 Specifications
Mains voltage indication
Display range
30V-270V±5V
Accuracy
Protective conductor resistance test
Measuring range
0.00-20.00Ω
Open circuit voltage 5V±0.4V DCMeasuring current
200mA DC(nominal value)±3%rdg±5dgt
Accuracy
Insulation resistance test
Rated volatge
250V500V
Measuring range
0.00-20.00MΩ
No-load voltage
250V DC +20%,-0%500V DC +20%,-0%
Short circuit current 1.5mA DC or less
±2%rdg±3dgt
Accuracy
Load current/Leakage current test
Item
Load current
Leakage current
Mains voltage range 100-253V/50Hz
Measuring range
0.10-10.00A rms
0.10-20.00mA rms±10%rdg±5dgt±3%rdg±5dgt
Accuracy
RCD test
Rated voltage
230V -15% to +10%/50Hz
Rated current
10mA/30mA×1×5
Function
Test duration
0.0ms-500.0ms
0.0ms-40.0ms
Operating time accuracy ±2ms(≦40ms),±8ms(>40ms)
Power source
LR6(AA)(1.5V) × 6
IEC/EN61010-1 CAT II 300V, IEC/EN61010-2-030,
Applicable Standards
IEC/EN61010-031, EN61326-2-2, AS/NZ3760
261(L) × 104(W) × 57(D)mm /Dimentions / WeightApprox. 930g(including batteries)
7277(Mains lead), 7129A(Test lead with Alligator clip),
7161A(Flat test prod), 7276(Adaptor for Extension cord),
Included Accessories
9193(Carrying case), 8928(Fuse[10A/250V]), 9121(Shoulder strap), Buckle, LR6(AA) x 6, Instruction manual
8263-USB(USB cable with "KEW Report(software)"),Optional Accessories 7275(Printer cable:Mini Din 6pin-D-sub 9pin）,
7248(Test lead with Alligator clip and Flat test probe)● Included Accessories

Function

Tests of contents
 Protective conductor resistance(Test current 200mA DC nominal)

ClassⅠTest

 Insulation resistance test (250V or 500V)
 Leakage current test (100-253V/50Hz)
 Load current test (100-253V/50Hz)
 Insulation resistance test (250V or 500V)
 Leakage current test (100-253V/50Hz)
 Load current test (100-253V/50Hz)
 Protective conductor resistanceClass Ⅱ Test

(Test current 200mA DC nominal)

Extension Lead Test

 Insulation resistance test

(between Line/Neutral-Earth short, Line/Neutral) Leakage current test (100-253V/50Hz)
 Polarity test
 RCD test (10mA/30mA)

RCD Test● Colour status back light
PASS / FAIL result complying with AS/NZS 3760

PASS

FAIL

● Connector panel

Special
4mm connector
mains lead for Class I earth
connection continuity andcan be 100- Class II insula253Vtion test leads

IEC connector for
IEC leads
and extension leads

USB terminal MINI DIN
for connecting terminal forto PC
connecting
to printer● Optional Accessories

7275

8263-USB"KEW Report"
Software for report“KEW Report” transfers
measurement data fromthe KEW6205 to a PC via
MODEL8263-USB.

System requirements
7248
7277

7129A

7161A

7276

9193

OS:
Display:
Hard-disk:
Others:Windows® Vista/7/8/10
XGA (Resolution 1024 × 768 dots) or moreSpace required 20Mbyte or more
With CD-ROM drive and USB port

* Windows® is a registered trademark of Microsoft in the United States.

Please read the "Safety Warnings" in the instruction manual supplied with the instrument thoroughly and completelycorrect use. Failure to follow the safety rules can cause fire, trouble, electrical shock, etc. Therefore, make sure
Safety Warnings : forto operate the instrument on a correct power supply and voltage rating marked on each instrument.

For inquires or orders :2-5-20, Nakane, Meguro-ku, Tokyo, 152-0031 Japan
Phone:+81-3-3723-0131
Fax:+81-3-3723-0152
E-mail:info-eng@kew-ltd.co.jp

The contents of this leaflet are subject to change without notice.

KEW 6205-1E Mar. 18 AD