Genius SQ 24

Nitrogen for Single Quad LC-MS
Part Number : 99-9999

Service Kit : 99-9999

Your local gas generation partner

Description
For labs using any Single Quad LC-MS systems, the Genius SQ
24 nitrogen gas generator delivers a high purity nitrogen supply
dedicated for this instrument. With a compact size to fit undermost lab benches and flow rates of up to 24 L/min, meeting andexceeding the maximum flow rate required by any Single QuadLC-MS on the market today, your lab can enjoy hassle free nitrogengas which helps you deliver the best analytical results.
Genius SQ 24 has been designed using Peak’s experience of overtwo decades of working with and developing the best nitrogengenerator membranes using expert suppliers and highly skilled
Peak design and manufacturing engineers. This rigorous design
process has helped not only deliver a reliable, high performancegas generator, it is also the best value nitrogen generator on themarket today.

Applications))

Single Quad LC-MS

Key Features))Best value nitrogen generator on the market))

Variable flow up to 24 l/min))Analytical grade nitrogen supply for Single Quad LCMS))

Outlet pressure up to 116psi (at 22 l/min maximum)))

Reduced height and compact size to fit under mostlab benches))

Integrated internal compressors for easy plug and play))

Based on proven and highly reliable technology))

Backed by Peak Protected global on-site servicing,technical support and partner training

750 mm / 29.6”

610 mm / 24.1”

600 mm / 23.4”

Copyright © Peak Scientific Instruments Ltd 2019 — Genius SQ 24 Data sheet Rev 5 RSID 2604 EN —17/12/2019

Technical Specifications
Voltage

Genius SQ 24100V

120V

Max Flow230V

Up to 24 L/min

Min/ Max PressureUp to 116 psi (@ 22 l/min max)

Gas outlets

1 x 1/4” BSPP

Max Relative Humidity

80% Non-Condensing

Max Altitude

2000m

Particles<0.01μm

Phthalates

NONE

Suspended liquids

NONE

Operating Temperature

5°C (41°F) to 35°C (95°F)

Electrical Requirements

50/60Hz, 11A

60Hz, 10A

50/60Hz, 6A

Power Consumption<1,100VA

1,200VA<1,380VA

Heat Output BTU

2934 @ 50Hz
3446 @ 60Hz

3685 @ 60Hz

5016 @ 50Hz
4231 @ 60Hz

610 x 600 x 750 mm
24.0 x 23.6 x 29.5’’

Generator Dimensions (H x W x D)
Generator Weight

83 kg / 183 lbs

Shipping Weight

104 kg / 229.3 lbs

Ordering Information
Part Number

3301661 (230v) | 3301660 (120v) | 3301650 (100v)

Annual Service
Complete Maintenance Plan†visit: www.peakscientific.com/ordering

Noise level expressed as SPL (Sound Pressure Level) measured at 1m from source in a reverberant chamber

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Peak Scientific’s Quality
Management System
conforms to:
ISO 9001:2015

Product Certifications

For a full list of our worldwide
office locations, please visit:www.peakscientific.com/offices
Web: www.peakscientific.com
Email: discover@peakscientific.com

Copyright 2019 Peak Scientific - Genius SQ 24 Data sheet Rev 5 RSID 2604 EN - 17/12/19