Precision Nitrogen
Nitrogen Generator for GC
Part Number : See Reverse

Service Kit : 08-3612

Your local gas generation partner

Description
Precision Nitrogen has been developed to provide a constantand consistent source of Nitrogen for detector make-up gas atstandard/typical detection levels for GC applications, as well asheadspace vial pressurization, purge and trap, tube conditioning
for thermal desorption and sample blow down. These generators
are capable of delivering high purity nitrogen, removing oxygen andmoisture via Pressure Swing Adsorption and Carbon Molecular Sieve
technology.
As with all Precision Series generators, Nitrogen models benefit
from a compact and modular, stackable design, minimizing the total
footprint required for GC gas supply, and providing flexibility to add
or remove modules as your laboratory requirements evolve over time.

Applications)

Sample Prep
• Headspace• Purge and Trap
•Tube Conditioning)

Detector Gas• Make-up / ReferenceKey Features540 mm)

Suitable as make-up gas at standard detection limits)

Nitrogen generated on demand, as and when it isrequired)Constant and consistent supply ,avoid running out ofgas during analysis)

Avoid contaminants entering system as result ofchanging over empty cylinders)

Dedicated Precision compressor module available
where house supply not available)

Integrates seamlessly with other Precision units)

Ultra-fast start-up time, quick to reach standard
operating purity)

Minimum maintenance with an annual filter change)

Compact, space-saving modular design

380 mm

256 mmPrecision Nitrogenshown in a typical Precision stack.

Copyright 2015 Peak Scientific - Precision Nitrogen Standard Data Sheet Rev 6 RSID 384 EN - 10/09/24

Technical Specifications

Nitrogen, 250cc

Nitrogen, 600ccNitrogen, 1L

250cc/min

600cc/min

1,000cc/min

Max Flow Rate
Max Pressure

80 psi / 5.5 bar

Purity> 99.9995%

Gas Outlets

1 x 1/4 " BSPP

Min/Max Air Inlet Pressure

6.9-8.3 bar / 100-120 psi

Min Air Inlet Flow35 lpm

Min Inlet Air Quality

ISO8573 - 1:2010 Class 1.4.1

Phthalates

None

Suspended Liquids

None

Start-Up Time For Purity

1.5 hours

Operating Temperature

5°C - 35°C / 41°F - 95 °F

Electrical Requirements

110 VAC ± 10% 60 Hz 0.37A / 230 VAC ± 10% 50 Hz 0.17A

Power Consumption

39.1 - 40.7 Watts

Heat Output

Up to 140 BTU/Hr

Generator Dimensions (WxDxH)
Generator Weight38.0 x 54 x 25.6 cm / 14.9 x 21.2 x 10”
21 Kg (46.2 lbs)

21 Kg (46.2 lbs)

26 Kg (57.3 lbs)

61-0250

61-0600

61-1000

Ordering Information
Part Number
Annual Service
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan

Precision Air Compressor

110v

230v

Annual Service Kit

Part Number

65-1555

65-0555

08-8372

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s QualityManagement System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2015 Peak Scientific - Precision Nitrogen Standard Data Sheet Rev 6 RSID 384 EN - 10/09/24