The Surgical Video Recorder,

Perfected

UR-X

Designed with Surgeons in mind

Plug & Play
DVI/HDMI, SDI

One-touch
Made inJapan

Recording5”

Built-in Touch
Display

Quality you can trust

1080p
60

DICOM
integration
c-store, MWM, MPPS,
Q/R, Offline Mode,
DICOM for Videoetc.

Operating Room

Capture / Record

Applications

Storage/Share

SurgicalLight Camera

Foot Switch

SurgicalMicroscope

DICOM

External Display

moduleHD/SD Images

RemoteC-STOREPACS Server
PC/ Mac

TabletEHR / EMR
UltrasoundWorklist

Video&Audio

Laparoscopy

HIS/RIS Server

Network
Network Transfer

Endoscopy

Patient Info

Live Recording

FTP/CIFS ServerCMSPC/ Mac

Content Management

AX-Angiography
USB
KeyboardiHDD

External Media

Product Overview

Key Features

Now, medical video recording is more relevant for all surgical ﬁelds. While it has become
more approachable than ever before, understanding all the many factors involved in thesetup can still be overwhelming. The variations in modality, camera system, hardware andsoftware can create a barrier to getting started. With TEAC's new UR-X all-in-one system,ease of use is as easy as 1-2-3, operability and connectivity are greatly enhanced without
adding complexity. Video Recording/ Transferring to any PACS and other features can beeasily and readily deployed to ﬁt your workﬂow and documentation needs. Having truly
dependable and high-quality solutions for medical video management, allowing surgeonsand clinicians to capture WHAT matters, WHEN it matters, is TEAC's mission.

Speciﬁcations

Real-time Recording x Medical Grade Full HD

Storage & Display
Recordable Devices

Internal Hard Drive (1TB), USB Flash Drive, USB Hard Drive

Built-in LCD Display

5-inch Touch Panel (Resistive type)

Recording Format

Compression

FileVideo

H.264 (Video), AAC (Audio)

mtsPhoto

JPEG, Uncompression

JPG, TIFF

Input/Output

Video

SDI (BNC)

3G-SDI, HD-SDI, SD-SDI

DVI *31920x1200*4, 1920x1080 (1080-60p/59.94i), 1280x 9 60, 1024x768

OtherInterfaces

More Integration x DICOM/PACS
Multiple Connectivity’s x Flexible Recording Possibilities

Recording Times (for a 1TB Hard Drive) *Analog HDTV (DVI-I *3)

1920x1080 (1080-59.9 4i )RGB (DVI-I * )525-60i

Stereo mini

MIC (Input only), LINE

20 Mbps

15 Mbps

10 Mbps

5 Mbps

RS-232C

D-sub 9 pin

111 hrs

148 hrs

222 hrs

444 hrs

3AudioOne-touch operation x Built-in Touch Screen
Plug & Play operation x Easy as 1-2-3

LANRJ-45 (1000BASE-T, 100BASE-TX, 10BASE-T)

USB3.0

USB Type-A x 2 on front (for Recordable Devices)

USB2.0

USB Type-A x 4 on rear (for Accessories)

External contact switch

Stereo Mini jack x 3

High-Deﬁnition 1920 x 1080p 60fps* Times are approximate.

GeneralPower supply

AC 100V - 230V, 50/60Hz

Dimensions (W x H x D)

Approx. 202 x 114 x 337 mm / 8.0 x 4.5 x 13.3 inch

WeightApprox. 3.7 kg / 8.2 lbs.
Operating temperature range 5°C to 40°C / 41°F to 104°F

Environments Operating humidity range

30 to 80% RH (no condensation)

Storage temperature range –10°C to 60°C / 14°F to 140°F
Certiﬁcations / Compliance

ANSI/AAMI ES60601-1, CAN/CSA-C22.2 NO.60601, CE MDR
IEC60601-1, EN60601-1, EN60601-1-2, FCC Class A,
FDA Class 1, RoHS

Accessories (Optional)

Foot-switch, Keyboard*1 Input/output terminal is used common to DVI-I. *2 1920 x1200 video recording is resized to 1728×1080.

Company names, product names and logos in this document are the trademarks or registered trademarks of their respective owners. * Features and speciﬁcations are subject to change without notice.

TEAC EUROPE GmbH.
Bahnstrasse 12, 65205
Wiesbaden-Erbenheim, Germany
Tel.: +49 -(0)611-7158-349
Fax: +49 -(0)611-7158-392
Email: info@teac.eu
https://teac.eu© Copyright TEAC CORPORATION 2021 2108 TCJ-PDF / ISD-121D