R

User Guide
Elcometer 456
Coating Thickness Gauge

www.elcometer.com

R

CONTENTS
Section

Page

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18

Gauge Overview
Box Contents
Using the Gauge
Getting Started
Connecting the Probe
Taking a Reading
Calibrating the Gauge
Locking & Unlocking the Calibration
Calibration Methods
Measurement Modes
Batching
Displaying Graphs
Menu Structure - Model T
Menu Structure - Model S
Menu Structure - Model B
Downloading Data & Upgrading Your Gauge
Technical Specification
Legal Notices & Regulatory Information

en-2
en-2
en-3
en-3
en-4
en-4
en-5
en-5
en-6
en-7
en-8
en-8
en-10
en-11
en-12
en-13
en-14
en-15

Pb

Hg

For the avoidance of doubt, please refer to the original English language version.
The Elcometer 456 is available in 3 models. This User Guide is written for the Model T. Where applicable theModel B and Model S are referenced.
Gauge Dimensions:
141 x 73 x 37mm (5.55 x 2.87 x 1.46")
Gauge Weight:
Integral: 156g (5.5oz) including batteries; Separate: 161g (5.68oz) including batteries.
Applicable patents:
US6243 661; US5886522; US6762603; US7606671; GB2306009; GB2367135;
GB2342450, DE10131827© Elcometer Limited 2012 - 2018. All rights reserved. No part of this document may be reproduced, transmitted,
transcribed, stored (in a retrieval system or otherwise) or translated into any language, in any form or by anymeans (electronic, mechanical, magnetic, optical, manual or otherwise) without the prior written permission of
Elcometer Limited.

en-1

www.elcometer.com

R

1 GAUGE OVERVIEW
6

1
2

7
3
8

4

5

1
2
3
4
5
6
7
8

LED Indicators - Red (left), Green (right)
Colour Screen
Multifunction Softkeys
On/Off Key
Internal Probe / Separate Probe Connection
USB Data Output Socket (below cover)
Battery Compartment (¼ turn open/close)
Wrist Strap Connection

2 BOX CONTENTS
Elcometer 456 Coating Thickness Gauge
Calibration Foils (Integral gauges)
Wrist HarnessProtective Case (B, S & T models)Transit Case (T model)

1 x Screen Protector (S & T models)
2 x AA Batteries

USB Cable & ElcoMaster® Software (S & T)
Test CertificateUser Guide
www.elcometer.com


en-2

R

3 USING THE GAUGEabBattery Life Indicator
Bluetooth On -

Model
o
BST
ST

n

Grey: not paired; Orange: paired

c
d
e
f
g

Substrate Type - F, N, FNF
Calibration Method
Reading Outside Calibration
Warning On
Upper Limit On
Units of Measurement -

BST
BST
T

m
l

ST
BST

k

µm, mils, mm, inchhBatch Type -

a
b
c
d
e
f
g
h
i
j
p

ST

normal, counted average, IMO

i
j
k
l
m
n

Menu Softkey
Display Softkey
Batch / Data Softkey
Calibration Softkey
Reading Value
User Selectable Statistics -

BST
BST
BST
BST
BST
BST

q

r

4 rowsopBatch Name (when in Batching)T
Date & Time (when not in batching) ST
Bar Graph - highest, lowest &BSTsaverage readingqrLower Limit On
Measurement Mode - Standard,

ST
T

t

Auto Repeat, Scanst

Run Chart - last 20 readings
Upper & Lower Limits On

ST
ST

4 GETTING STARTED
4.1 ENSURING YOUR GAUGE HAS THE LATEST FIRMWARE &UPGRADING YOUR GAUGETo ensure that your gauge has the most up-to-date gauge firmware,
allowing you to benefit from the latest features and functionality, we®
recommend that the gauge is connected to ElcoMaster on a regularbasis and before first use.
Simply connect the gauge via USB to an internet connected computer®
running ElcoMaster using the ‘Connect Gauge’ feature. If a later
version of the gauge firmware is available, ‘Update Gauge’ will be
displayed to the right of the gauge details. Click ‘Update Gauge’ toinstall the latest firmware.
www.elcometer.com
en-3

R

4 GETTING STARTED (continued)
4.2 SELECTING YOUR LANGUAGE
1
Press and hold the ON/OFF button until the Elcometer logo isdisplayed.
2
Select your language using the  softkeys.
3
Follow the on screen menus.
To access the language menu when in a foreign language:1
Switch the gauge OFF.
2
Press and hold the left softkey and switch the gauge ON.
3
Select your language using the  softkeys.
5 CONNECTING YOUR PROBESEPARATE GAUGES ONLY
1
Rotate the probe plug to align the pins.
2
Screw in the collar - clockwise.

2

1

2

1

6 TAKING A READING
1
2
3


•
•
•

Hold the probe by its sleeve.
To take a reading, bring the probe down onto the surface whilst
holding it perpendicular.
For subsequent readings, lift the probe off and then replace it on
to the coated surface.
DO
Hold the probe by the probe sleeve.
Gently place the probe onto the surface.
Allow the sleeve to make contact with the surface - to improveaccuracy.

www.elcometer.com

en-4

R

6 TAKING A READING (continued)
 DO NOT•Drag the probe over the coated surface.
•
Bang the probe down hard onto the surface.
•
Allow the probe to hover over the surface as this could result ina false reading.
The display will dim if inactive for more than 15 seconds and will go ‘black’ ifinactive for the period defined in Menu/Setup/Screen Settings/Screen Timeout.
Either press any key, or tap the gauge to awaken it.
The gauge will switch off automatically after 5 minutes of inactivity.
- - - indicates reading outside range of probe.

7 CALIBRATING THE GAUGE
1
2
3
4



Press the Cal softkey.
For alternative calibration methods,
select Cal/Cal Method.
Select Calibrate and follow the onscreen instructions.
When prompted place the probe on
the centre of the foil(s).

90°

Foils can be stacked.
Not all calibration methods are available for all gauge types see 9 Calibration
Methods.

8 LOCKING AND UNLOCKING THE CALIBRATIONTo set a calibration lock PIN code:1
Press the Menu softkey and selectSetup/Calibration Lock.
2
Set the four digit PIN Code using the softkeys to select 0 to 9
and the  softkey to move from thefirst to the fourth digit†.
3
Press Ok, Escape to cancel or Adjust
to amend the PIN code.

†

The  softkey will appear when the “ X “ changes to a number.

en-5

www.elcometer.com

R

8 LOCKING AND UNLOCKING THE CALIBRATION (continued)
To unlock the calibration:
1
Press the Cal softkey and select Calibration Lock.
2
Enter the four digit PIN Code, if set, using the  softkeys toselect 0 to 9 and the  softkey to move from the first to the fourth
digit†.
3
Press Ok or Escape to cancel.
To disable the calibration lock PIN Code:1
Press the Menu softkey and select Setup/Calibration Lock.
2
Enter the four digit PIN Code.


Should the user forget or loose the PIN code, the PIN can be disabled via®
ElcoMaster . Using the USB cable, simply connect the gauge to a PC with®
ElcoMaster Version 2.0.33 or higher installed and select Edit/Clear CalibrationPIN.

9 CALIBRATION METHODS†CALIBRATIONMETHODMODEL

DESCRIPTION

Zero

B, S, T

A one point calibration - ideal for smoothsurfaces, simply place the probe on theuncoated substrate and the gauge willautomatically calibrate

SmoothB, S, T

A two point calibration - the user selects a
suitable calibration foil and a smooth,uncoated substrate (zero)

Rough/ 2 Point

B, S, T

The ideal calibration method for rough or
profiled substrates, using two foils of known
thickness - one above, another below thetarget dry film thickness

Zero Offset

S, T

Calibration method for measurement when
the substrate profile/ roughness is unknown,or not accessible. A user defined offset value
is applied to the measurement reading

The  softkey will appear when the “ X “ changes to a number.

www.elcometer.com

en-6

R

9 CALIBRATION METHODS (continued)
CALIBRATIONMETHODMODEL

DESCRIPTION

Auto

S, T

Ideal for repetitive inspections. The user
stores known calibration foil thicknesses into
the gauge. When calibrating the user is
prompted to calibrate the gauge and thegauge automatically adjusts the readings to
the stored foil thicknesses - simplifying andconsiderably speeding up the calibration
processISO

S, T

Sets the calibration method to Zero Offset
and sets the Counted Average to 5 - inaccordance with ISO19840

SSPC PA2

S, T

Sets the calibration method to Rough/2
Point and sets the Counted Average to
3 - in accordance with SSPC PA2

SwedishS, T

Sets the calibration method to Rough/2
Point and sets the Counted Average to
5 - in accordance with SS standards

AustralianS, T

Sets the calibration method to Zero Offset,
and sets the Counted Average to 5 - inaccordance with AS standards

10 MEASUREMENT MODES (MODEL T)
10.1 MEASUREMENT MODES
Three measurement modes are available: “Standard Mode”, “AutoRepeat Mode” and “Scan Mode”.
To select the measurement mode, press Menu/Setup/Measurement Mode.
“Auto Repeat Mode” and “Scan Mode” are only available with an Elcometer 456
Ultra/Scan Probe. For further information contact Elcometer or visit
www.elcometer.com.

10.2 NAVSEA MODE
Displays the reading to the nearest 1mil (1µm) above 10mils (254µm)
and 0.1mil (0.1µm) below 10mils (254µm).
To select NAVSEA mode, press Menu/Setup/NAVSEA or Batch/New
Batch/NAVSEA.
en-7

www.elcometer.com

R

11 BATCHING (MODEL S & T)
1
To use the Batching memory facility, press the Batch softkey.
2
Select “New Batch” or “Open Existing Batch” to add readings.
3
Copy and review batch data.
4
Select “Edit Batch” to rename, clear readings from or delete abatch.
5
Fixed Batch Size allows users to pre-define the number ofreadings to be stored in a batch. Once all readings have beentaken the gauge automatically opens a new batch with a linkto the original batch name. For Example NewBatch_1 becomes
NewBatch_2, NewBatch_3, etc.


Save each reading into memory or store the average of a pre-defined number ofreadings using the Counted Average function.

12 DISPLAYING GRAPHS
12.1 BAR GRAPH
The Bar Graph displays an analogue
representation of the thickness value
together with the highest, lowest andaverage reading as measurements aretaken. To display the Bar Graph:
1

Press the Display softkey and select“Readings & Bar Graph”.



If a reading is outside set limits, the white bar and the reading value turn red.
When in “Scan” Mode (Model T), the bar graph is automatically displayed during
each scan.

12.2 RUN CHART (MODEL S & T)
To display the Run Chart of the last 20
readings:
1
2
3
Press the Batch softkey.
Select “New Batch” or “Open
Existing Batch”.
Press the Display softkey and select“Readings & Run Chart”.
Red points signify a reading outside the batch’s limits (if set).

www.elcometer.com

en-8

R

12 DISPLAYING GRAPHS (continued)
12.3 BATCH GRAPH (MODEL T)
To display the Batch Graph:
1
2



en-9

Select the appropriate batch namefrom Batch/Review Batch.
Select “Batch Graph”.
Red columns signify a reading outside thebatch’s limits (if set).
Press the Zoom+ softkey followed by  or 
to review individual readings as required.

www.elcometer.com

Select Statistics
All PSPC Values
Number of Readings
Mean
Lowest Reading
Highest ReadingStandard Deviation
Coefficient of Variation
Low Limit Value
Number Below Limit
High Limit Value
Number Above Limit
Elcometer Index ValueNDFT Value%>=NDFT% 90-100% NDFTIMO 90:10 Pass/Fail

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

www.elcometer.com
○
○
○
○
○

where n = 1, 2, or 3

Cal Memory n
Saved hh:mm dd/mm/yy
SelectCalibrate
Cal Method
Test Calibration
Rename Cal Memory nView Calibration Data
Clear Cal Memory n

ZeroSmoothRough / 2 Point
Zero Offset
Auto

Cal Method

Statistics
Clear Statistics
View SelectedView All
Select Statistics
Counted Average

Calibration
Calibrate
Cal Method
Cal Memory 1
Cal Memory 2
Cal Memory 3
Factory Calibration
Calibration Lock
Test Calibration
Calibration Check

Batch
Memory Free
New Batch
Open Existing Batch
Exit Batching
Edit Batch
Copy Batch
Review Batch
Deleted Reading

Display
Clear Statistics
Readings Only
Readings & Selected Stats
Readings & Run Chart
Readings & Bar Graph
Batch Graph
Statistics

Menu
Delete Last Reading
Setup
Set Limits
Bluetooth
About
Reset
Emergency Light

○

○

○
○
○
○

○

○

○

Deleted Reading
Delete With Tag
Delete Without Tag

Statistics
Batch Information
Calibration Information
Readings
Batch Graph

Review Batch
Batch 1

Edit Batch
Rename Batch
Clear Batch
Delete Batch

New Batch
Open Batch 1
Rename Batch 1
Batch Calibration
Batch Type
Batch Limits
Fixed Batch Size
NAVSEA

About
Gauge Information
Probe Information
Contact
Legal

○
○

○

BluetoothEnable Bluetooth○Bluetooth ID
xxxxxxBluetooth CodexxxxxxAI

Set Limits
Enable Limits
Set Upper
Set Lower
Set NDFT

Units
Language
Time and Date
Beep Volume
Screen Settings
Probe
Measurement Mode
Gauge Auto Off
Calibration Lock
NAVSEA

Setup

Units

○
○
○
○

Probe
Auto FNF
Ferrous F
Non Ferrous N
Probe Switch On

Batch Limits
Enable Limits
Set Upper
Set Lower
Set NDFT

Batch Type
NormalCounted Average
IMO PSPC

Batch Calibration
Calibrate
Cal Method
Use Cal Memory 1
Use Cal Memory 2
Use Cal Memory 3

Legal
Patents
Legal Notices
Regulatory

○

○
○
○

○
○
○

○
○
○
○

Measurement Mode
Standard Mode
Auto Repeat Mode
Scan Mode
Scan Mode With Hold
Calibration Lock
Enable PIN○○

Ferrous F1
Ferrous F2

Probe

○
○

○

○
○
○
○

Screen Settings
Screen Brightness
Screen Timeout
Auto Display Rotation
Opening Screen

Time and Date
Set Time
Set Date
Set Format
Display Time and Date

µm
mm
mils
inch

Probe

Regulatory

Cal Method
Zero
SmoothRough / 2 Point
Zero Offset
Auto
ISO
SSPC PA2
SwedishAustralian

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Menu displayed is dependent
on the probe connectedR

13 MENU STRUCTURE - ELCOMETER 456 MODEL T

en-10

en-11

Select Statistics
All PSPC Values
Number of Readings
Mean
Lowest Reading
Highest ReadingStandard Deviation
Coefficient of Variation
Low Limit Value
Number Below Limit
High Limit Value
Number Above Limit
Elcometer Index ValueNDFT Value%>=NDFT% 90-100% NDFTIMO 90:10 Pass/Fail

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

Zero
SmoothRough / 2 Point
Zero Offset
Auto

Cal Method

Statistics
Clear Statistics
View SelectedView All
Select Statistics
Counted Average

○
○
○
○
○
Calibration
Calibrate
Cal Method
Factory Calibration
Calibration Lock
Test Calibration

Batch
Memory Free
New Batch
Open Existing Batch
Exit Batching
Edit Batch
Review Batch
Deleted Reading

Display
Clear Statistics
Readings Only
Readings & Selected Stats
Readings & Run Chart
Readings & Bar Graph
Statistics

Menu
Delete Last Reading
Setup
Set Limits
Bluetooth
About
Reset
Emergency Light

○

○
○
○
○

○

○

Deleted Reading
Delete With Tag
Delete Without Tag

Statistics
Batch Information
Calibration Information
Readings

Review Batch
Batch 1

Edit Batch
Clear Batch
Delete Batch

New Batch
Open Batch
Batch Calibration
Batch Type

About
Gauge Information
Probe Information
Contact
Legal○○

BluetoothEnable Bluetooth○Bluetooth ID
xxxxxxBluetooth CodexxxxxxAI

Set Limits
Enable Limits
Set Upper
Set Lower
Set NDFT

Setup
Units
Language
Time and Date
Beep Volume
Screen Settings
Probe
Gauge Auto Off
Calibration Lock

○
○
○
○

Probe
Auto FNF
Ferrous F
Non Ferrous N
Probe Switch On

Batch Type
NormalCounted Average
IMO PSPC

Batch Calibration
Calibrate
Cal Method

Legal
Patents
Legal Notices
Regulatory

Calibration Lock
Enable PIN

○
○
○

○
○

○
○

Screen Settings
Screen Brightness
Screen Timeout
Auto Display Rotation
Opening Screen

Probe

○

Time and Date
Set Time
Set Date
Set Format
Display Time and Date

Ferrous F1
Ferrous F2

○
○
○
○

Units
µm
mm
mils
inch

Probe

Cal Method
Zero
SmoothRough / 2 Point
Zero Offset
Auto
ISO
SSPC PA2
SwedishAustralian

QDID

Regulatory

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Menu displayed is dependent
on the probe connectedR

14 MENU STRUCTURE - ELCOMETER 456 MODEL S

www.elcometer.com

www.elcometer.com

Select Statistics
Number of Readings
Mean
Lowest Reading
Highest ReadingStandard Deviation
Coefficient of Variation
Elcometer Index Value

○
○
○
○
○
○
○

Zero
SmoothRough / 2 Point

Cal Method

Statistics
Clear Statistics
View SelectedView All
Select Statistics
Counted Average○○○

Calibration

Delete51.5µmCalibrate
Cal Method
Factory Calibration
Calibration Lock
Test Calibration

EscapeData
Last Five Readings
50.5 µm52.3 µm51.0 µm51.8 µm

Display
Clear Statistics
Readings Only
Readings & Selected Stats
Readings & Bar Graph
Statistics

Menu
Delete Last Reading
Setup
About
Reset
Emergency Light

○

Ok

○
○
○

About
Gauge Information
Probe Information
Contact
Legal

Setup
Units
Language
Beep Volume
Screen Settings
Probe
Gauge Auto Off
Calibration Lock

○

Probe

Patents
Legal Notices
Regulatory

Legal

Calibration Lock
Enable PIN

Ferrous F1
Ferrous F2

○
○

○
○
○
○

○
○

Screen Settings
Screen Brightness
Screen Timeout
Auto Display Rotation
Opening Screen
Probe
Auto FNF
Ferrous F
Non Ferrous N
Probe Switch On

○
○
○
○

Units
µm
mm
mils
inch

Probe

Regulatory
xxxxx

Menu displayed is dependent
on the probe connectedR

15 MENU STRUCTURE - ELCOMETER 456 MODEL B

en-12

R

16 DOWNLOADING DATA & UPGRADING YOUR GAUGE
16.1 USING ELCOMASTER®
Using ElcoMaster® - supplied with the Elcometer 456 Model S & T, andavailable as a free download at elcometer.com - all gauges cantransmit readings to a PC for archiving and report generation. Data
can be transferred via USB or Bluetooth® (Models S & T). For more
information on ElcoMaster® visit www.elcometer.com.
16.2 USING ELCOMASTER® MOBILE APPS (MODEL S & T)
Ideal when out in the field or on-site, using the ElcoMaster® Android™
or iOS Mobile App users can:






Store live readings directly on to amobile device and save them into
batches together with GPS
coordinates.
Add photographs of the test surface.
Map readings on to a map,
photograph or diagram.
Inspection data can be transferredfrom mobile to PC for further analysis
and reporting.

For more information on ElcoMaster®
Mobile Apps visit www.elcometer.com

Compatible with smart phones
and tablets running Android 2.1
or above. To install, download
via www.elcometer.com or using
the Google Play™ Store app,and follow the on screen
instructions.

en-13

www.elcometer.com

R

16 DOWNLOADING DATA & UPGRADING YOUR GAUGE (cont.)
Made for iPhone 6 Plus, iPhone 6,
iPhone 5s, iPhone 5c, iPhone 5,
iPhone 4s, iPhone 4, iPad Air 2,
iPad mini 3, iPad Air, iPad mini 2,
iPad (3rd and 4th generation), iPadmini, iPad 2, and iPod touch (4thand 5th generation). To install,
download via www.elcometer.com
or the App Store, and follow the on
screen instructions.
16.3 UPGRADING YOUR GAUGE
Gauge firmware can be upgraded to the latest version by the user via
ElcoMaster®, as it becomes available. ElcoMaster® will inform the userof any updates when the gauge is connected to the PC with aninternet connection.

17 TECHNICAL SPECIFICATION
BatteryType
OperatingTemperature

2 x AA batteries, rechargeable batteries
can also be used
-10 to 50°C (14 to 122°F)

Relative
Humidity

0 to 95%

Gauge
Dimensions

14.1 x 7.30 x 3.70cm(5.55 x 2.87 x 1.46")

Gauge Weight

Integral: 156g (5.5oz)
Separate: 161g (5.68oz)(with supplied
batteries)

Can be used in accordance with:
AS 2331.1.4, AS 3894.3-B, AS/NZS 1580.108.1, ASTM B 499, ASTM D 1186-B,
ASTM D 1400, ASTM D 7091, ASTM E 376, ASTM G 12, BS 3900-C5-6B,
BS 3900-C5-6A, BS 5411-11, BS 5411-3, BS 5599, DIN 50981, DIN 50984,
ECCA T1, EN 13523-1, IMO MSC.215(82), IMO MSC.244 (83), ISO 1461,
ISO 19840, ISO 2063, ISO 2178, ISO 2360, ISO 2808-6A, ISO 2808-6B,
ISO 2808-7C, ISO 2808-7D, ISO 2808-12, NF T30-124, SS 184159, SSPC PA 2,
US Navy PPI 63101-000, US Navy NSI 009-32

www.elcometer.com

en-14

R

18 LEGAL NOTICES & REGULATORY INFORMATION
Declaration of Conformity:
Elcometer 456 Model B complies with the requirements of the following EU Directives:
2014/30/EU
Electromagnetic Compatibility
2011/65/EU
Restriction of the use of certain hazardous substances
Elcometer 456 Models S & T comply with the requirements of the following EU Directives:
2014/53/EU
Radio Equipment
2011/65/EU
Restriction of the use of certain hazardous substances
Declarations of Conformity are available to download via:
Model B: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/English/DoC_456C_B.pdf
Models S & T: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/English/DoC_456C_S&T.pdf
Operational Frequency Band: 2,402 - 2,480 MHz
Maximum Transmitted Power: <4 dBm
This product is Class B, Group 1 ISM equipment according to CISPR 11.
Class B product: Suitable for use in domestic establishments and in establishments directly connected to a low voltage
power supply network which supplies buildings used for domestic purposes.
Group 1 ISM product: A product in which there is intentionally generated and/or used conductively coupled radio frequency
energy which is necessary for the internal functioning of the equipment itself.
The USB is for data transfer only and is not to be connected to the mains via a USB mains adapter.
This device complies with Part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) this device may
not cause harmful interference, and (2) this device must accept any interference received, including interference that may
cause undesired operation.
The ACMA compliance mark can be accessed via: Menu/About/Legal/Regulatory
Elcometer 456 Models S & T: The Giteki mark, its product identification code, the FCC ID and Bluetooth SIG QDID can beaccessed via: Menu/About/Legal/Regulatory
NOTE: This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of
the FCC Rules. These limits are designed to provide reasonable protection against harmful interference in a residential
installation. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used inaccordance with the instructions, may cause harmful interference to radio communications. However, there is no guaranteethat interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or
television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct theinterference by one or more of the following measures:-- Reorient or relocate the receiving antenna.
-- Increase the separation between the equipment and receiver.
-- Connect the equipment into an outlet on a circuit different from that to which the receiver is connected-- Consult the dealer or an experienced radio/TV technician for help.
To satisfy FCC RF Exposure requirements for mobile and base station transmission devices, a separation distance of 20 cm
or more should be maintained between the antenna of this device and persons during operation. To ensure compliance,
operation at closer than this distance is not recommended. The antenna(s) used for this transmitter must not be co-located or
operating in conjunction with any other antenna or transmitter.
Modifications not expressly approved by Elcometer Limited could void the user’s authority to operate the equipment under
FCC rules.
This device complies with Industry Canada license exempt RSS standard(s). Operation is subject to the following two
conditions: (1) this device may not cause interference, and (2) this device must accept any interference, including interference
that may cause undesired operation of the device.
This Class B digital apparatus complies with CAN ICES-3 (B)/NMB-3(B).
and ElcoMaster® are registered trademarks of Elcometer Limited, Edge Lane, Manchester,M43 6BU. United Kingdom
are trademarks owned by Bluetooth SIG Inc and licensed to Elcometer Limited.
Elcometer 456 Models S & T: Made for iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPhone 4s, iPhone 4, iPad
Air 2, iPad mini 3, iPad Air, iPad mini 2, iPad (3rd and 4th generation), iPad mini, iPad 2, and iPod touch (4th and 5th
generation).
“Made for iPod,” ”Made for iPhone,” and “Made for iPad” mean that an electronic accessory has been designed to connect
specifically to iPod, iPhone, or iPad, respectively, and has been certified by the developer to meet Apple performance standards.
Apple is not responsible for the operation of this device or its compliance with safety and regulatory standards. Please note thatthe use of this accessory with iPod, iPhone, or iPad may affect wireless performance.
iPad, iPhone, and iPod touch are trademarks of Apple Inc., registered in the U.S. and other countries.
App Store is a trademark of Apple Inc., registered in the U.S. and other countries.
Google Play is a trademark of Google Inc.
All other trademarks acknowledged.
R

The Elcometer 456 is packed in a cardboard package. Please ensure that all packaging is disposed of in an environmentallysensitive manner. Consult your local Environmental Authority for further guidance.
Head-Office: Elcometer Limited, Edge Lane, Manchester, M43 6BU, United Kingdom.

TMA-0488 Issue 17 - Text with cover 22373

R

Guide d'utilisation
Elcometer 456
Jauge d'épaisseur de revêtement

www.elcometer.com

R

TABLE DES MATIERES
Section1
Présentation de la jauge
2Contenu de la boîte
3
Utiliser la jauge
4Premières démarrer
5
Brancher la sonde
6
Prendre une mesure7
Étalonner la jauge
8
Bloquer et débloquer la calibration
9
Méthodes d'étalonnage
10Modes de mesure
Enregistrement en lots
11
Afficher les graphiques
1213Structure du Menu - Modèle T
14
Structure du Menu - Modèle S
15
Structure du Menu - Modèle B
16
Transférer les données & Actualiser votre jauge
17
Caractéristiques techniques
18
Mentions légales et dispositions règlementaires

Pb

Page
fr-2
fr-2
fr-3
fr-3
fr-4
fr-4
fr-5
fr-5
fr-6
fr-7
fr-8
fr-8
fr-10
fr-11
fr-12
fr-13
fr-14
fr-15

Hg

Pour éviter tout doute, référez-vous à la version anglaise originale.
L'Elcometer 456 est disponible en 3 versions. Ce mode d'emploi décrit l’utilisation du modèle T. Le cas échéant lemodèle B et modèle S sont référencés.
Dimensions de la jauge :141 x 73 x 37mm (5.55 x 2.87 x 1.46")
Poids de la jauge :Intégrée : 156 g (5.5oz) batteries incluses;
Séparée : 161g (5.68oz) batteries incluses.
Brevets applicables :US6243 661; US5886522; US6762603; US7606671; Gb2306009;
GB2367135; GB2342450, DE10131827© Elcometer Limited 2012 - 2018. Tous droits réservés. Aucune partie de ce document peut être reproduite,transmise, stockée (dans un système d'extraction ou autre), ou traduite quelque langue que ce soit, sous quelqueforme ou moyen que ce soit (électronique, mécanique, magnétique, optique ou autre), sans l'autorisation préalableécrite d'Elcometer Limited.

fr-1

www.elcometer.com

R

1 PRÉSENTATION DE LA JAUGE
6

1
2

7
3
8

4

5

1
2
3
4
5
6
7
8

LED indicatrices - Rouge (gauche), Verte (droite)
Ecran couleur
Boutons multifonctions
Bouton On/Off
Sonde intégrée / Connexion pour sonde séparée
Sortie de données USB (sous le cache)Compartiment piles (¼ de tour ouverture / fermeture)
Crochet pour cordon de poignet

2 CONTENU DE LA BOÎTE











Jauge d'épaisseur de revêtement Elcometer 456
Feuilles d'étalonnage (jauges à sonde intégrée)
Cordon de poignet
Etui protecteur (modèles B, S & T)
Valise de transport (modèle T)
1 x Protecteur d'écran (modèles S & T)
2 x piles AA
Câble USB & logiciel ElcoMaster® (S & T)
Certificat de test
Guide d'utilisationwww.elcometer.com

fr-2

R

3 UTILISER LA JAUGE
Modèleab

Indicateur de niveau de batterie BST
ST
Bluetooth activé Gris : non appairé ; Orange : appairé

c
d
e
f
g
h

BST
Type de subjectile - F, N, FNF
BST
Méthode d'étalonnage
T
Alarme de mesure horsétalonnage activée
ST
Limite maximum activée
Unité de mesure - µm, mils, mm, inch BST
ST
Type de lot -

a

o

b
c
n

d
e
f
g

m

h

l

i
j

k

p

normal, moyenne, IMO

i
j
k
l
m
n

Touche Menu
Touche Afficha
Touche Lot / Mesure
Touche d'étalonnage
Valeur de la mesure
Statistiques sélectionnables -

BST
BST
BST
BST
BST
BST

q

r

4 lignesopTNom du lot (en mode lots)
Date et heure (hors du mode lots) ST
Graphique barres - Mesure maxi, BSTsmini & moyenneqrST
Limite minimum activée
Mode de Mesure - Standard, Auto Tt

Repeat, ScanstGraphe - 20 dernières mesures
Limites maximum et minimum
activées

ST
ST

4 PREMIÈRES DÉMARRER
4.1

GARANTIR QUE VOTRE JAUGE COMPORTE LE DERNIERFIRMWARE & MISE À JOUR DE VOTRE JAUGE
Pour vous assurer que votre jauge possède le firmware le plus à jour, afinde bénéficier des dernières caractéristiques et fonctionnalités, nous®
recommandons de connecter votre jauge à ElcoMaster régulièrement etavant la première utilisation.
Il suffit de connecter la jauge via USB à un ordinateur connecté à Internet®
avec ElcoMaster en utilisant la fonction ‘Connecter Jauge’. Si une version
plus récente du firmware est disponible pour la jauge, le message 'Mise àjour jauge’ sera affiché à droite des détails de la jauge. Cliquez sur ‘Mise àjour jauge’ pour installer le dernier firmware.

fr-3

www.elcometer.com

R

4 PREMIÈRES DÉMARRER (suite)
4.2 SELECTION DE LA LANGUE
1Maintenez la touche ON/OFF appuyée jusqu'à ce que le logo
Elcometer apparaisse.
2
Sélectionnez votre langue à l'aide des touches .
3
Suivez les instructions du menu à l'écran.
Pour accéder au menu langues lorsque la jauge est en langue étrangère:
1
Eteignez la jauge.
2
Maintenez la touche gauche appuyée et allumez la jauge.
3
Sélectionnez votre langue à l'aide des touches .
5 BRANCHER LA SONDEJAUGES A SONDE SÉPARÉE UNIQUEMENT
1
Tournez la sonde pour l'aligner avec la prise.
2
Vissez l'embout dans le sens des aiguilles d'une montre.

2

1

2

1

6 PRENDRE UNE MESURE
1
2
3

Tenez la sonde par le manchon.
Pour prendre une mesure, positionnez la sonde vers le bas sur lasurface tout en la maintenant perpendiculaire.
Pour les mesures successives, soulevez la sonde, puis replacezla sur la surface revêtue.



FAIRE
Tenir la sonde par le manchon.
••
Placez délicatement la sonde sur la surface.
•
Laissez le manchon entrer en contact avec la surface - pouraméliorer la précision.
www.elcometer.com

fr-4

R

6 PRENDRE UNE MESURE (suite) NE PAS FAIRE•••

Faire glisser la sonde sur la surface peinte.
Frapper brusquement la surface avec la sonde.
Survoler la surface avec la sonde, car cela pourrait entraîner
une fausse lecture.

L'intensité de l'écran diminue s'il n'est pas utilisé pendant 15 secondes, etdevient 'noir' au bout de la période déterminée dans Menu/Initialiser/Réglagesécran/Temps Écran Dépassé. Appuyez sur une touche ou tapotez légèrement lajauge pour l'allumer.
La jauge s'éteint automatiquement après 5 minutes d'inactivité.
- - - indique les mesures hors de l'échelle de la sonde.



7 ÉTALONNER LA JAUGE
1
2
3
4



Pressez le bouton Cal.
Pour une autre méthode decalibration, sélectionnez Cal/MéthodeCal.
Sélectionnez Étalonner et suivez les
instructions a l'écran.
Lorsque vous y êtes invité, placez lasonde au centre de la/les feuille(s).

90°

Les feuilles d'étalonnage peuvent être superposées.
Toutes les méthodes d'étalonnage ne sont pas disponibles pour tous les types dejauges. Voir paragraphe 9 Méthodes d’Étalonnage.

8 BLOQUER ET DÉBLOQUER LA CALIBRATION
Pour définir un code PIN de verrouillage :1
Appuyez sur la touche Menu etsélectionnez Setup/Verrouillage Cal.
2
Pour entrer le code PIN à 4 chiffres,utilisez les flêches  pour faire
un choix entre 0 et 9, puis la flèche
 pour vous déplacer du premierau quatrième chiffre†.
3
Appuyez sur OK pour confirmer, sur "Echap" pour annuler ou sur"Ajust" pour modifier le code PIN.

†

La flêche  apparait lorsque le “ X “ se change en chiffre.

fr-5

www.elcometer.com

R

8 BLOQUER ET DÉBLOQUER LA CALIBRATION (suite)
Pour déverrouiller la calibration :1
Appuyez sur la touche CAL et sélectionnez Verrouillage Cal.
2
Tapez le code PIN à 4 chiffres, à l'aide des flêches  pour faire
un choix entre 0 et 9, puis de la flèche  pour vous déplacer dupremier au quatrième chiffre†.
3
Appuyez sur OK pour confirmer ou "Echap" pour annuler.
Pour désactiver le code PIN de verrouillage :1
Appuyez sur la touche Menu et sélectionnez Setup/
Verrouillage Cal.
2
Tapez le code PIN à 4 chiffres.
®

En cas de perte ou d'oubli, le code PIN peut être désactivé via ElcoMaster . A
l'aide du câble USB, connectez la jauge à un PC équipé de la version
ElcoMaster® 2.0.33 ou supérieure, puis sélectionnez Edit/Effacer PIN calibration.

9 MÉTHODES D'ÉTALONNAGE†

MÉTHODE
D'ÉTALONNAGEMODÈLEDESCRIPTION

Zéro

B, S, TUn étalonnage en un point - idéal pour lessurfaces lisses, il suffit de placer la sonde sur lesubstrat sans revêtement et la jauge s'étalonne
automatiquement

Lisse

B, S, TUn étalonnage en deux points - l'utilisateursélectionne une feuille d'étalonnage appropriéeet un substrat lisse, sans revêtement (zéro)

Rugueuse/en 2 points

B, S, T

La méthode d'étalonnage idéale pour les
substrats rugueux ou profilés, à l'aide de deux
feuilles d'épaisseur connue - l'une d'une valeur
supérieure à l'épaisseur du revêtement à
mesurer, et l'autre d'une valeur inférieure

Compensation
zéro

S, T

Méthode d'étalonnage pour mesurer lorsque leprofil du substrat ou sa rugosité est inconnue
ou n'est pas accessible. Une valeur decompensation définie par l'utilisateur est
appliquée à la valeur de la mesure

La flêche  apparait lorsque le “ X “ se change en chiffre.

www.elcometer.com

fr-6

R

9 MÉTHODES D'ÉTALONNAGE (suite)MÉTHODE
D'ÉTALONNAGEMODÈLEDESCRIPTION

Auto

S, T

Idéal pour des inspections répétées.
L'utilisateur rentre des épaisseurs de feuilles
d'étalonnage connues dans la jauge. Lors del'étalonnage, l'utilisateur est invité à étalonner lajauge et la jauge ajuste automatiquement lesvaleurs de l'épaisseur des feuilles rentrées
dans la jauge auparavant - simplifie et accélère
considérablement le processus d'étalonnage

ISO

S, T

Sélectionne la méthode d'étalonnage
Compensation Zéro et définit les moyennes,
calculées sur 5 mesures - conformément à lanorme ISO19840

SSPC PA2

S, T

Sélectionne la méthode d'étalonnage
Rugueuse/ en 2 points et définit les moyennes,
calculées sur 3 mesures - conformément à lanorme SSPC PA2

SuédoisS, T

Sélectionne la méthode d'étalonnage
Rugueuse/ en 2 points et définit les moyennes,
calculées sur 5 mesures - conformément aux
normes SS

Australien

S, T

Sélectionne la méthode d'étalonnage
Compensation Zéro et définit les
moyennes, calculées sur 5 mesures conformément aux normes AS

10 MODES DE MESURE (MODÈLE T)
10.1 MODES DE MESURE
Il existe trois modes de mesure : ‘Mode Standard’, ‘Mode Auto Repeat’et ‘Mode Scan’.
Pour choisir le mode de mesure, appuyez sur Menu/Initialiser/Mode Mesure.
Les modes ‘Auto Repeat’ et ‘Scan’ sont uniquement accessibles avec une sonde
Ultra/Scan pour Elcometer 456. Pour en savoir plus, contactez Elcometer ou
visitez notre site sur www.elcometer.com.

10.2 MODE NAVSEA
Affiche la mesure au 1mil (1µm) le plus proche au-dessus de 10mils(254μm) et 0,1mil (0,1 µm) en-dessous de 10mils (254μm).
Pour choisir le mode NAVSEA, appuyez sur Menu/Initialiser/NAVSEA
ou Lot/Nouveau Lot/NAVSEA.
www.elcometer.com
fr-7

R

11 ENREGISTREMENT EN LOTS (MODÈLES S ET T)
1
2
3
4
5



Pour utiliser la fonction Enregistrement en lots, pressez le bouton
Lots.
Sélectionnez ‘Nouveau Lot’ ou ‘Ouvrir Le Lot Existant’ pour
ajouter des mesures.
Copiez ou révisez le lot.
Sélectionnez ‘Editer le lot’ pour renommer un lot, effacer lesmesures ou un lot entier.
La Taille Fixe du Lot permet aux utilisateurs de prédéfinir lenombre de mesures à stocker dans un lot. Une fois que toutes lesmesures ont été prises, la jauge ouvre automatiquement un
nouveau lot avec un lien vers le nom du lot d'origine. Par exemple
NewBatch_1 devient NewBatch_2, NewBatch_3, etc.
Enregistrez chaque mesure dans la mémoire ou stockez la moyenne d'unnombre prédéfini de mesures en utilisant la fonction Moyenne Calculée.

12 AFFICHER LES GRAPHIQUES
12.1 GRAPHIQUE BARRES
Le graphique en barres donne unereprésentation de l'épaisseur ainsi que lavaleur maxi, mini et moyenne en temps réel.
Pour afficher le Graphique en barres :1

Appuyez sur la touche Afficha, puissélectionnez ‘Données & Graph. barres’.

Si une valeur est en dehors des limites définies, la barre blanche et la valeur lue
deviennent rouge.
En mode ‘Scan’ (modèle T), le graphique en barres s'affiche automatiquement àchaque scan.



12.2 GRAPHE (MODÈLES S ET T)
Pour afficher le graphe des 20 dernières
mesures :123
Pressez le bouton Lots.
Sélectionnez ‘Nouveau Lot’ ou‘Ouvrir Le Lot Existant’.
Appuyez sur la touche Afficha, puissélectionnez ‘Données & Graphe’.
Les points rouges indiquent une mesure en dehors des limites du lot (si définies).

www.elcometer.com

fr-8

R

12 AFFICHER LES GRAPHIQUES
12.3 GRAPHIQUE LOT (MODÈLE T)
Pour afficher le graphe relatif au lot
de mesures :
1
2



fr-9

Sélectionnez le nom du lot souhaité
dans Lot/Réviser Le Lot.
Sélectionnez ‘Graphique Lot’.
Les colonnes rouges indiquent une valeur en dehors des limites du lot(si définies).
Appuyez sur la touche Zoom+ suivie de  ou  pour revoir les valeurs
individuelles si vous le souhaitez.

www.elcometer.com

www.elcometer.com

Sélectionner Stats.
Toutes Valeurs PSPC
Nombre De Mesures
MoyenneMes. La Plus Basse
Mes. La Plus Haute
Ecart-Type
Coeff. De Variation
Valeur Limite Basse
Chiffre Inf. à La Limite
Valeur Limite Haute
Chiffre Sup. à La Limite
EIV
Valeur NDFT%>=NDFT% 90-100% NDFTIMO 90:10 Réussi/Raté

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

où n = 1, 2, ou 3

Mémoire Cal. nVide hh:mm dd/mm/yy
Sélect.
Calibrer
Méthode Cal.
Tester La Calibration
Renomm. Mémoire Cal. n
Voir Les Données D’Calibration
Effacer Mémoire Cal. n

Méthode Cal.
Zéro
Lisse
Rug./ 2 points
Remise à Zéro
Auto

Statistiques
Supprimer Stats.
Voir Sélection
Tout Voir
Sélectionner Stats.
Moyenne Calculée

Calibration
Calibrer
Méthode Cal.
Mémoire Cal. 1
Mémoire Cal. 2
Mémoire Cal. 3
Calibration Usine
Verrouillage Cal
Tester La Calibration
Vérifier La Calibration

Lot
Mémoire Disponible
Nouveau Lot
Ouvrir Le Lot ExistantSortir Du Lot
Editer Le Lot
Copier Le Lot
Réviser Le Lot
Mesure Effacée

AffichaSupprimer Stats.
Données uniquement
Données & stats sélection
Données & Graphe
Données & Graph. barres
Graphique Lot
Statistiques

Menu
Suppr. Dernière Mesure
Initialiser
Fixer Les Limites
BluetoothAu Sujet De...
Ré-Initialiser
Eclairage D'urgence

○

○

○
○
○
○

○

Mesure Effacée
Suppr. Avec Mém.
Suppr. Sans Mém.

Réviser Le Lot
Lot 1
Statistiques
Informations Sur Le Lot
Informations Calibration
Mesures
Graphique Lot

Editer Le Lot
Renommer Le Lot
Effacer Le Lot
Supprimer Le Lot

Nouveau Lot
Ouvrir Lot 1
Renomm. Lot 1
Calibration Du Lot
Type De Lot
Limites Du Lot
Taille De Lot Fixe
NAVSEA

Au Sujet De...
Information Jauge
Information Sonde
Adresses Elcometer
Légal

○
○

○

Bluetooth
Autoriser Bluetooth○
ID Bluetooth
xxxxxx
Code Bluetooth
xxxxxxAI

Fixer Les Limites
Autoriser Lim.
Fixer Lim. Haute
Fixer Lim. Basse
Fixer NDFT

UnitésLangueHeure et Date
Bip Volume
Réglages écran
Sonde
Mode Mesures
Extinction automatique jauge ○Verrouillage Cal○
NAVSEA

Initialiser

○

○
○
○
○
○
○

Heure et Date
Màj L'heure
Màj La Date
Màj La Date
Affich. Heure et Date
Réglages écran
Brilliance De l'écran
Temps Écran Dépassé
Rotation Autom. ÉcranOuverture de l'écran
Sonde
Auto FNF
Ferreux F
Non-Ferreux N
Mise en route via sonde

Limites Du Lot
Autoriser Lim.
Fixer Lim. Haute
Fixer Lim. Basse
Fixer NDFT

Type De Lot
NormalMoyenne Comptée
IMO PSPC

Calibration Du Lot
Calibrer
Méthode Cal.
Utilis. Mémoire Cal. 1
Utilis. Mémoire Cal. 2
Utilis. Mémoire Cal. 3

Brevets
Informations Légales
Réglementation

Légal

○

○
○
○

○
○
○

○
○
○
○

Mode Mesures
Mode Standard
Mode Auto Repeat
Mode Scan
Mode Scan + maintenir
Verrouillage Cal
Utiliser un code PIN○○

Ferreux F1
Ferreux F2

Sonde

○
○
○
○

Unités
µm
mm
mils
inch

Réglementation

Méthode Cal.
Zéro
Lisse
Rug./ 2 points
Remise à Zéro
Auto
ISO
SSPC PA2
SuédoisAustralien

QDID

Sonde

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Le menu affiché dépend
de la sonde connectéeR

13 STRUCTURE DU MENU - ELCOMETER 456 MODÈLE T

fr-10

fr-11

Sélectionner Stats.
Toutes Valeurs PSPC
Nombre De Mesures
MoyenneMes. La Plus Basse
Mes. La Plus Haute
Ecart-Type
Coeff. De Variation
Valeur Limite Basse
Chiffre Inf. à La Limite
Valeur Limite Haute
Chiffre Sup. à La Limite
EIV
Valeur NDFT%>=NDFT% 90-100% NDFTIMO 90:10 Réussi/Raté

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

Méthode Cal.
Zéro
Lisse
Rug./ 2 points
Remise à Zéro
Auto

Statistiques
Supprimer Stats.
Voir Sélection
Tout Voir
Sélectionner Stats.
Moyenne Calculée

○
○
○
○
○
Calibrer
Méthode Cal.
Calibration Usine
Verrouillage Cal
Tester La Calibration

Calibration

Lot
Mémoire Disponible
Nouveau Lot
Ouvrir Le Lot ExistantSortir Du Lot
Editer Le Lot
Réviser Le Lot
Mesure Effacée

AffichaSupprimer Stats.
Données uniquement
Données & stats sélection
Données & Graphe
Données & Graph. barres
Statistiques

Menu
Suppr. Dernière Mesure
Initialiser
Fixer Les Limites
BluetoothAu Sujet De...
Ré-Initialiser
Eclairage D'urgence

○

○
○
○
○

Mesure Effacée
Suppr. Avec Mém.
Suppr. Sans Mém.

Réviser Le Lot
Lot 1
Statistiques
Informations Sur Le Lot
Informations Calibration
Mesures

Editer Le Lot
Effacer Le Lot
Supprimer Le Lot

Nouveau Lot
Ouvrir Lot
Calibration Du Lot
Type De Lot

Au Sujet De...
Information Jauge
Information Sonde
Adresses Elcometer
Légal

○
○

Bluetooth
Autoriser Bluetooth○
ID Bluetooth
xxxxxx
Code Bluetooth
xxxxxxAI

Type De Lot
NormalMoyenne Comptée
IMO PSPC

Calibration Du Lot
Calibrer
Méthode Cal.

Légal
Brevets
Informations Légales
Réglementation

Verrouillage Cal
Utiliser un code PIN

○
○
○

○
○

○
○
○
○

Sonde
Auto FNF
Ferreux F
Non-Ferreux N
Mise en route via sonde

Fixer Les Limites
Autoriser Lim.
Fixer Lim. Haute
Fixer Lim. Basse
Fixer NDFT

Sonde

○
○

Réglages écran
Brilliance De l'écran
Temps Écran Dépassé
Rotation Autom. ÉcranOuverture de l'écran

Ferreux F1
Ferreux F2○Heure et Date
Màj L'heure
Màj La Date
Màj La Date
Affich. Heure et Date

UnitésLangueHeure et Date
Bip Volume
Réglages écran
Sonde
Extinction automatique jauge ○Verrouillage Cal

○

○
○
○
○

Initialiser

Unités
µm
mm
mils
inch

Sonde

Réglementation

Méthode Cal.
Zéro
Lisse
Rug./ 2 points
Remise à Zéro
Auto
ISO
SSPC PA2
SuédoisAustralien

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Le menu affiché dépend
de la sonde connectéeR

14 STRUCTURE DU MENU - ELCOMETER 456 MODÈLE S

www.elcometer.com

www.elcometer.com

Sélectionner Stats.
Nombre De Mesures
MoyenneMes. La Plus Basse
Mes. La Plus Haute
Ecart-Type
Coeff. De Variation
EIV

○
○
○
○
○
○
○

Méthode Cal.
Zéro
Lisse
Rug./ 2 points

Statistiques
Supprimer Stats.
Voir Sélection
Tout Voir
Sélectionner Stats.
Moyenne Calculée○○○

Calibration

Effacer

Calibrer
Méthode Cal.
Calibration Usine
Verrouillage Cal
Tester La Calibration

Echap.

51.5µmData
5 Dernières Mesures
50.5 µm52.3 µm51.0 µm51.8 µm

AffichaSupprimer Stats.
Données uniquement
Données & stats sélection
Données & Graph. barres
Statistiques

Menu
Suppr. Dernière Mesure
Initialiser
Au Sujet De...
Ré-Initialiser
Eclairage D'urgence

○

Ok

○
○
○

Au Sujet De...
Information Jauge
Information Sonde
Adresses Elcometer
Légal

UnitésLangueBip Volume
Réglages écran
Sonde
Extinction automatique jauge ○Verrouillage Cal

Initialiser

Brevets
Informations Légales
Réglementation

Légal

Verrouillage Cal
Utiliser un code PIN

○
○

○
○
○
○

Sonde
Auto FNF
Ferreux F
Non-Ferreux N
Mise en route via sonde
Sonde

○
○

Réglages écran
Brilliance De l'écran
Temps Écran Dépassé
Rotation Autom. ÉcranOuverture de l'écran

Ferreux F1
Ferreux F2

○
○
○
○

Unités
µm
mm
mils
inch

Sonde

Réglementation
xxxxx

Le menu affiché dépend
de la sonde connectéeR

15 STRUCTURE DU MENU - ELCOMETER 456 MODÈLE B

fr-12

R

16 TRANSFÉRER LES DONNÉES & ACTUALISER VOTRE JAUGE
16.1 AVEC LE LOGICIEL ELCOMASTER®
Utiliser ElcoMaster® - fourni avec les modèles S et T del'Elcometer 456, et disponible en téléchargement gratuit sur le site
elcometer.com - toutes les jauges peuvent transmettre des mesures à
un PC pour l'archivage et la création de rapports. Les données
peuvent être transférées via USB ou Bluetooth® (modèles S & T). Pourplus d'informations sur ElcoMaster®, visitez www.elcometer.com.
16.2 AVEC LE LOGICIEL ELCOMASTER® MOBILE APPS(MODÈLES S & T)
Idéal lorsque vous êtes sur site ou en chantier ; grâce au logiciel
ElcoMaster® Android™ ou iOS Mobile App, vous pouvez :






Stocker les données en temps réel sur
votre mobile et les enregistrer dans deslots avec les coordonnées GPS.
Ajouter des photos de la zone de test.
Cartographier les mesures sur une carte,une photo ou un diagramme.
Transférer les données d'inspection devotre mobile vers un PC pour analyseultérieure et édition de rapports.

Pour en savoir plus sur ElcoMaster® Mobile
Apps, visitez notre site www.elcometer.com

APPLI ANDROID SUR

fr-13

Compatible avec smartphones ettablettes équipés d'Android
version 2.1 ou suivantes. Pourl'installation, téléchargez lelogiciel via www.elcometer.com
ou utilisez Google Play™ Storeapp, et suivez les instructions àl'écran.

www.elcometer.com

R

16 TRANSFÉRER LES DONNÉES & ACTUALISER VOTRE JAUGE (suite)
Conçu pour iPhone 6 Plus,
iPhone 6, iPhone 5s, iPhone 5c,
iPhone 5, iPhone 4s, iPhone 4,
iPad Air 2, iPad mini 3, iPad Air,
iPad mini 2, iPad (3ème et 4èmegénération), iPad mini, iPad 2, etiPod touch (4ème et 5ème
génération). Pour l'installation,
téléchargez le logiciel via
www.elcometer.com ou utilisez
App Store, et suivez les
instructions à l'écran.
16.3 ACTUALISER VOTRE JAUGE
Vous pouvez actualiser le logiciel interne de votre jauge avec ladernière version disponible via ElcoMaster®. ElcoMaster® vous informedès qu'une mise à jour est disponible lorsque votre jauge est
connectée à un PC équipé d'une connexion Internet.
17 CARACTÉRISTIQUES TECHNIQUESType de piles

2x piles AA, des piles rechargeables peuventégalement être utiliséesTempérature defonctionnementde -10 à 50°C (de 14 à 122°F)

Humidité relativede 0 à 95%

Dimensions de lajauge

14,1 x 7,30 x 3,70cm(5,55 x 2,87 x 1,46")

Poids de la jauge

Intégrée: 156g (5,5oz)
Séparée: 161g (5,68oz)(piles inclues)

Conforme aux normes:
AS 2331.1.4, AS 3894.3-B, AS/NZS 1580.108.1, ASTM B 499, ASTM D 1186-B,
ASTM D 1400, ASTM D 7091, ASTM E 376, ASTM G 12, BS 3900-C5-6B,
BS 3900-C5-6A, BS 5411-11, BS 5411-3, BS 5599, DIN 50981, DIN 50984,
ECCA T1, EN 13523-1, IMO MSC.215(82), IMO MSC.244 (83), ISO 1461,
ISO 19840, ISO 2063, ISO 2178, ISO 2360, ISO 2808-6A, ISO 2808-6B,
ISO 2808-7C, ISO 2808-7D, ISO 2808-12, NF T30-124, SS 184159, SSPC PA 2,
US Navy PPI 63101-000, US Navy NSI 009-32

www.elcometer.com

fr-14

R

18 MENTIONS LÉGALES ET DISPOSITIONS RÈGLEMENTAIRES
Déclaration de Conformité :L’Elcometer 456 modèle B réponds aux exigences des directives UE suivantes :2014/30/EU
Compatibilité électromagnétiques
2011/65/EU
Limitation de l’utilisation de certaines substances dangereuses
L’Elcometer 456 modèles S et T répondes aux exigences des directives UE suivantes :2014/53/EU
Équipements radioélectriques
2011/65/EU
Limitation de l’utilisation de certaines substances dangereuses
Les déclarations de conformité sont disponibles au téléchargement à :Modèle B: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/French/DoC_456C_B.pdf
Modèles S et T: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/French/DoC_456C_S&T.pdf
Bande de fréquence opérationnelle : 2.402 - 2.480 MHz
Puissance maximale émise : <4 dBm
Ce produit est un équipement de Classe B, Groupe 1 ISM conformément au CISPR 11.
Les produits de Classe B peuvent être utilisés dans les établissements domestiques et dans les établissements directement
reliés à un réseau basse tension qui alimente des bâtiments à usage domestique.
Produit ISM de Groupe 1 : produit dans lequel on génère et/ou utilise intentionnellement l’énergie radioélectrique nécessaire
au fonctionnement interne de l’équipement lui-même.
La prise USB est uniquement destinée au transfert des données et ne doit pas être branchée sur le secteur via un adaptateur
USB/Secteur.
Cet appareil est conforme a la partie 15 des normes FCC. Son utilisation est sujette aux deux conditions suivantes: (1) Cet
appareil ne doit pas générer d'interférences, et (2) cet appareil doit accepter toute interférence reçue, dont les interférencesqui pourraient causer un fonctionnement indésiré.
Vous pouvez avoir accès à la marque de conformité ACMA via : Menu/Au Sujet De../Légal/Réglementation
Elcometer 456 Modèles S & T: La marque Giteki, son numéro d'ordonnance, le FCC ID et le SIG QDID Bluetooth sontaccessibles dans : Menu/Au Sujet De../Légal/Réglementation
NOTE: cet appareil a été teste et a été déclaré conforme aux limites imposées pour un appareil numérique de Classe B,
conformément à la Partie 15 des règlements de la FCC. Ces limites sont conçues pour fournir une protection raisonnable contre lesinterférences nuisibles dans une installation résidentielle. Cet équipement génère, utilise et peut émettre des fréquences radio et, s'iln'est pas installé et utilisé conformément aux instructions, peut causer des interférences nuisibles aux communications radio.
Cependant, il n'existe aucune garantie que des interférences ne se produiront pas dans une installation donnée. Si cet équipement
provoque des interférences nuisibles à la réception radio ou de télévision, ce qui peut être déterminé en mettant l'équipement horstension; l'utilisateur est encouragé à essayer de corriger l'interférence par une ou plusieurs des mesures suivantes:-- Réorientez ou déplacez l'antenne réceptrice.
-- Augmentez la distance entre l'appareil et le récepteur.
-- Branchez l'appareil dans une prise sur un circuit différent de celui auquel le récepteur est connecté.
-- Consultez votre revendeur ou un technicien radio / TV expérimenté.
Pour satisfaire aux exigences de la FCC relatives à l'exposition aux radiofréquences (RF) pour les appareils de transmission
mobiles et les stations de base, il faut garder une distance de séparation de 20 cm ou plus entre l'antenne de cet appareil etles personnes pendant l'utilisation. Pour garantir la conformité, nous déconseillons d'utiliser l'appareil à une distance inférieureà celle-ci. La ou les antenne(s) utilisée(s) pour cet émetteur ne doivent pas être installée(s) ou utilisée(s) en conjonction avec
d'autres antennes ou émetteurs.
Les modifications non expressément approuvées par Elcometer Limited peuvent annuler l'autorisation de l'utilisateur d'utiliser
cet appareil selon les règles de la FCC.
Le présent appareil est conforme aux CNR d'Industrie Canada applicables aux appareils radio exempts de licence. L'exploitation
est autorisée aux deux conditions suivantes: (1) l'appareil ne doit pas produire de brouillage, et (2) l'utilisateur de l'appareil doit
accepter tout brouillage radioélectrique subi, même si le brouillage est susceptible d'en compromettre le fonctionnement.
Cet appareil numérique de classe B est conforme au CAN ICES-3 (B)/NMB-3(B).
et ElcoMaster® sont les marques déposée de Elcometer Limited, Edge Lane, Manchester,M43 6BU. Royaume-Uni.
est une marque détenue par Bluetooth SIG Inc et don't l'autorisation d'utilisation à été donnéeà Elcometer Limited.
Elcometer 456 Modèles S & T: Conçu pour iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPhone 4s, iPhone 4, iPad Air 2,
iPad mini 3, iPad Air, iPad mini 2, iPad (3ème et 4ème génération), iPad mini, iPad 2, et iPod touch (4ème et 5ème génération).
La mention “Made for iPod,” ”Made for iPhone,” et “Made for iPad” signifie qu'un accessoire électronique a été spécialement
conçu pour se connecter aux iPod, iPhone, ou iPad et a été certifié par le développeur pour répondre aux normes derendement Apple. Apple n'est pas responsable du fonctionnement de cet équipement ou de sa conformité aux normes desécurité et aux dispositions légales. Nous vous informons que l'utilisation de cet accessoire avec un iPod, iPhone, ou iPadpeut nuire aux performances de la liaison sans fil.
iPad, iPhone, et iPod touch sont des marques déposées d'Apple Inc., enregistrées aux U.S.A. et dans d'autres pays.
App Store est une marque déposée d'Apple Inc., enregistrée aux U.S.A. et dans d'autres pays.
Google Play est une marque déposée de Google Inc.
Toutes les autres marques sont reconnues.
R

L'Elcometer 456 est livré dans un emballage en carton. Merci d'éliminer tous les emballages de manière écologique.
Contactez la déchèterie de votre localité pour plus d'informations sur le recyclage.
Siège Social : Elcometer Limited, Edge Lane, Manchester, M43 6BU. Royaume Uni.

TMA-0488-01 Numéro 17 - Texte avec couverture 22373-01

R

Bedienungsanleitung
Elcometer 456
Schichtdickenmessgerät

www.elcometer.com

R

INHALTSVERZEICHNIS
Abschnitt

Seite

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18

Geräteüberblick
Verpackungsinhalt
Nutzung des Gerätes
InbetriebnahmeAnschluss der Sonde
Durchführung von Messungen
Kalibrierung des Gerätes
Setzen und Aufheben der Kalibriersperre
Kalibriermethoden
Messmodi
Loserstellung
Diagrammanzeige
Menüstruktur - Modell T
Menüstruktur - Modell S
Menüstruktur - Modell B
Datendownload und MessgerätupgradeTechnische Daten
Rechtliche Hinweise und behördliche Informationen

de-2
de-2
de-3
de-3
de-4
de-4
de-5
de-5
de-6
de-7
de-8
de-8
de-10
de-11
de-12
de-13
de-14
de-15

Pb

Hg

Im Zweifelsfall hat immer die Originalanleitung auf Englisch Vorrang.
Das Elcometer 456 gibt es in drei Ausführungen. Diese Anleitung bezieht sich auf das Modell T. Wo relevant
werden Verweisungen auf die Modelle B und S vorgenommen.
Geräteabmessungen:
140 x 720 x 450mm (5,51 x 2,83 x 1,77").
Gerätegewicht:
Integrierte Sonde: 156g (5,5oz) inklusive Batterien.
Separate Sonde: 161g (5,68oz) inklusive Batterien.
Anwendbare Patente:
US6243 661; US5886522; US6762603; US7606671; GB2306009;
GB2367135; GB2342450, DE10131827© Elcometer Limited 2012 - 2018. Alle Rechte vorbehalten. Kein Teil dieses Dokumentes darf ohne vorherige
schriftliche Genehmigung durch Elcometer Limited vervielfältigt, übertragen, gespeichert (gleich in welcher Art und
Weise) oder in eine andere Sprache übersetzt werden (gleich welcher Form: elektronisch, mechanisch,magnetisch, optisch oder in anderer Weise).

de-1

www.elcometer.com

R

1 GERÄTEÜBERBLICK
6

1
2

7
3
8

4

5

1
2
3
4
5
6
7
8

LED Anzeigen - Rot (links), Grün (rechts)
Farbbildschirm
Multifunktional programmierte Tasten
AN/AUS - Taste
Integrierte Sonde / Externer Sondenanschluss
USB Datenausgangsanschluss (unter dem Deckel)
Batteriefach (¼-Umdrehung zum Öffnen/Schliessen)
Handschlaufenverbindung

2 VERPACKUNGSINHALT











Elcometer 456 Schichtdickenmessgerät
Kalbrierfolien (bei Geräten mit integrierter Sonde)
Handschlaufe
Schutztasche (B, S & T Modelle)
Transportkoffer (T Modell)
1 x Bildschirmschutz (S & T Modell)2 x AA Batterien
USB Kabel & ElcoMaster® Software (S & T)
Testzertifikat
Bedienungsanleitung

www.elcometer.com

de-2

R

3 NUTZUNG DES GERÄTES
ab

Batteriestandsanzeige
Bluetooth AN - Grau: nicht

Modell o

a

BST
ST

b
c
n

gekoppelt; Orange: gekoppeltcUntergrundtyp - F (Eisen),

BST

N (Nicht-Eisen), FNF (Kombi)dKalibriermethodeeWarnung für Messwert außerhalb T
des Kalibrierbereichs AN

f
g
h

BST

ST
Oberer Grenzwert AN
Maßeinheiten - µm, mils, mm, inch BST
ST
Lostyp - normal, gezählter

m
l
k

d
e
f
g
h
i
j

p

Durchschnitt, IMO

i
j
k
l
m
n

Taste für Menü
Taste für Display
Taste für Los/Daten
Taste für Kalibrierung
Messwert
Benutzerdefinierte Statistik -

BST
BST
BST
BST
BST
BST

q

r

4 ReihenopTLosname (wenn im Losmodus)
Datum & Zeit (wenn nicht im Losmodus) ST
Balkengrafik - höchster, niedrigster BSTsund durchschnittlicher Messwertqr

Unterer Grenzwert AN
Mess-Modus - Standard-,

ST
T

t

Wiederholautomatik-, Scan-stST
Diagramm - letzte 20 Messwerte
Unterer & Oberer Grenzwert AN ST

4 INBETRIEBNAHME
4.1 WIE SIE IHRE MESSGERÄTEFIRMWARE AUF DEN NEUESTENSTAND BRINGEN UND HALTEN
Um sicherzustellen, dass die Firmware Ihres Messgerätes, d.h. die
Gerätesoftware auf dem neuesten Stand ist und sie damit auch dieneuesten Funktionen nutzen können, empfehlen wir, dass Sie Ihr
Messgerät vor der ersten Nutzung und danach immer wieder regelmäßig
an die ElcoMaster® Auswertungssoftware anschließen.

de-3

www.elcometer.com

R

4 INBETRIEBNAHME (Fortsetzung)Verbinden Sie dazu Ihr Messgerät über die USB Schnittstelle mit einem PC
auf den vorher die ElcoMaster® heruntergeladen wurde. Die Verbindunggeschieht über die Funktion ‚Gerät verbinden‘. Wenn es eine aktuellere
Gerätefirmware geben sollte, dann wird rechts von den
Geräteinformationen die Nachricht ‚Gerät updaten‘ erscheinen. Drücken
Sie dann ‚Gerät updaten‘, um die aktuellste Firmware in das Gerät zu
laden.
4.2 AUSWAHL IHRER SPRACHE
1
Drücken und halten Sie die AN/AUS Taste bis das Elcometer
Logo erscheint.
2
Wählen Sie Ihre Sprache durch Nutzung der  Tasten.
3
Folgen Sie den Anweisungen auf dem Bildschirm.
So kommen Sie aus einer fremden Sprache in die Sprachauswahl:
1
Schalten Sie das Gerät AUS.
2
Drücken und halten Sie die linke Taste und schalten das Gerät EIN.
3
Wählen Sie Ihre Sprache durch Nutzung der  Tasten.
5 ANSCHLUSS DER SONDENUR BEI GERÄTEN MIT SEPARATER SONDE
1
Drehen Sie den Sondenstecker, um die Kontakte miteinander
auszurichten.
2
Schrauben Sie den Verbindungsring im Uhrzeigersinn ein.

2

1

2

6 DURCHFÜHRUNG VON MESSUNGEN
1
2

1

Halten Sie die Sonde an der Sondenhülse.
Um eine Messung vorzunehmen setzen Sie die Sonde senkrecht
auf die Oberfläche.
3
Für aufeinanderfolgende Messungen nehmen Sie die Sonde vonder beschichteten Oberfläche und setzen Sie sie wieder auf.
de-4
www.elcometer.com

R

6 DURCHFÜHRUNG VON MESSUNGEN (Fortsetzung) BITTE WIE FOLGT VORGEHEN:
Halten Sie die Sonde an der Sondenhülse.
••Setzen Sie die Sonde sanft auf die Oberfläche auf.
•
Um die Genauigkeit zu erhöhen, setzen Sie die Sondenhülse
auf die Oberfläche auf.
 BITTE UNTERLASSEN SIE FOLGENDES:
Nie die Sonde über die beschichtete Oberfläche ziehen!
••
Nie die Sonde hart auf die Oberfläche aufsetzen!
•
Die Sonde nicht über die Oberfläche "schweben" lassen da dieszu falschen Messwerten führen kann.


Die Anzeige wird sich verdunkeln wenn das Gerät für mehr als 15 Sekunden
nicht benutzt wird und wird "schwarz" wenn das Gerät für eine unter
Menü/Einstellung/Bildschirmeinstellungen/Abschaltzeit Bildschirm eingestellte
Zeitdauer inaktiv ist. Drücken Sie eine beliebige Taste oder tippen Sie das
Messgerät an, um es zu aktivieren.
Das Gerät schaltet sich nach 5 Minuten Nicht-Benutzung automatisch aus.
- - - zeigt an, dass der Messwert außerhalb des Sondenmessbereichs liegt.

7 KALIBRIERUNG DES MESSGERÄTES
1
Drücken Sie die Taste "Kal".
2
Für andere Kalibriermethoden
drücken Sie auf Kal-methode.
3
Wählen Sie KALIBRIEREN undfolgen Sie den Anweisungen auf dem
Bildschirm.
4
Bei Aufforderung platzieren Sie die
Sonde im Zentrum der Folie(n).



90°

Folien können gestapelt werden.
Nicht alle Kalibriermethoden sind in allen Gerätetypen verfügbar (siehe
Abschnitt 9 unter Kalibriermethoden)

8 SETZEN UND AUFHEBEN DER KALIBRIERSPERRE
Festlegen des PIN-Codes zum Sperren
der Kalibrierung:1
Drücken Sie die Taste Menü undwählen Sie Einstellung/Kalibriersperre.
2
Setzen Sie den vierstelligen PIN-Code
mit den Tasten  zur Auswahl von 0
bis 9 und der Taste  zur Auswahl derersten bis zur vierten Stelle†.
3
Drücken Sie OK zum Bestätigen, Abbrechen zum Verwerfen derÄnderung oder Anpass. zum Ändern des PIN-Codes.
†

Die Taste  erscheint, wenn das “X“ durch eine Ziffer ersetzt wird.

de-5

www.elcometer.com

R

8 SETZEN UND AUFHEBEN DER KALIBRIERSPERRE (Fort.)
Aufheben der Kalibriersperre:1
Drücken Sie die Taste Kal und wählen Sie Kalibriersperre.
2
Geben Sie den vierstelligen PIN-Code, falls bereits festgelegt, mit
den Tasten  zur Auswahl von 0 bis 9 und der Taste  zur
Auswahl der ersten bis zur vierten Stelle ein†.
3
Drücken Sie OK oder Abbrechen.

Deaktivieren des PIN-Codes der Kalibriersperre:1
Drücken Sie die Taste Menü und wählen Sie Einstellung/
Kalibriersperre.
2
Geben Sie den vierstelligen PIN-Code ein.


Falls der PIN-Code vergessen oder verloren wurde, kann er mit ElcoMaster®
deaktiviert werden. Schließen Sie das Messgerät mit dem USB-Kabel an einen PC®
an, auf dem ElcoMaster Version 2.0.33 oder höher installiert ist und wählen Sie
Bearbeiten/Kalibrier-PIN löschen.

9 KALIBRIERMETHODEN†

KALIBRIERMETHODE

MODELLBESCHREIBUNG

Null

B, S, T

Bei der Ein-Punkt-Kalibrierung - ideal für glatte
Oberflächen - setzen Sie die Sonde einfach auf
den unbeschichteten Untergrund und das Gerät
wird automatisch kalibriert.

Glatt

B, S, T

Bei der Zwei-Punkt-Kalibrierung wählt derAnwender eine passende Kalibrierfolie und einenglatten, unbeschichteten Untergrund (Null).

Rau / ZweiPunkt

B, S, T

Die ideale Kalibriermethode für raue odergestrahlte Untergründe: Einsatz von zwei Folien
bekannter Stärke - eine über, eine unter derangestrebten Trockenfilmdicke.

Null-Offset

S, T

Eine Kalibriermethode für Messungen bei denen
das Profil/die Rauheit des Untergrundes
unbekannt oder unzugänglich ist. Ein vom
Anwender definierter Offset-Wert wird vom
Messwert abgezogen.

Die Taste  erscheint, wenn das “X“ durch eine Ziffer ersetzt wird.

www.elcometer.com

de-6

R

9 KALIBRIERMETHODEN (Fortsetzung)
KALIBRIERMETHODE

MODELLBESCHREIBUNG

Auto

S, T

Ideal für sich wiederholende Inspektionen. Der
Anwender speichert bekannte
Kalibrierfolienstärken in das Gerät. Bei der
Kalibrierung wird der Anwender aufgefordert,das Gerät zu kalibrieren und das Gerät passt
automatisch die Messwerte an die
gespeicherten Folienstärken an. Diesvereinfacht und beschleunigt den
Kalibrierprozess erheblich.

ISO

S, T

Setzt die Kalibriermethode auf Null-Offset und
setzt den Gezählten Durchschnitt auf 5 - inÜbereinstimmung mit ISO19840.

SSPC PA2

S, T

Setzt die Kalibriermethode auf Rau/Zwei-Punkt
und setzt den Gezählten Durchschnitt auf 3 - inÜbereinstimmung mit SSPC PA2.

SchwedischS, T

Setzt die Kalibriermethode auf Rau/Zwei-Punkt
und setzt den Gezählten Durchschnitt auf 5 - inÜbereinstimmung mit SS Standards.

AustralischS, T

Setzt die Kalibriermethode auf Null-Offset und
setzt den Gezählten Durchschnitt auf 5 - inÜbereinstimmung mit AS Standards.

10 MESSMODI (MODELL T)
10.1 MESSMODI
Es sind drei Messmodi verfügbar: "Standard-Modus","Wiederholautomatik" und "Scan-Modus".
Drücken Sie zur Auswahl des Messmodus Menü/Einstellung/Mess-Modus.
"Wiederholautomatik" und Scan-Modus" sind nur in Verbindung mit einer
Elcometer 456 Ultra/Scan-Sonde verfügbar. Weitere Informationen erhalten Sie
bei Elcometer oder www.elcometer.com.

10.2 NAVSEA MODUS
Zeigt den Messwert mit einer Genauigkeit von 1mil (1µm) über 10mils(254µm) und 0.1mil (0.1µm) unter 10mils (254µm).
Drücken Sie zur Auswahl des NAVSEA modus Menü/Einstellung/
NAVSEA oder Los/Neues Los/NAVSEA.
de-7

www.elcometer.com

R

11 LOS-ERSTELLUNG (MODELL S & T)
1
Um den LOS-Speicher zu nutzen, drücken Sie die Taste "LOS”.
2
Wählen Sie "Neues Los" oder “Öffne Vorhandenes Los" um
Messwerte hinzuzufügen.
3
Kopieren und überprüfen Sie die Losdaten.
4
Wählen Sie zum Umbenennen, zum Löschen von Messwerten
und zum Löschen eines Loses "Bearbeite Los”.
5
Die Funktion "Feste Losgröße" erlaubt dem Anwender, die Zahl
der in einem Los zu speichernden Messwerte vorzugeben. Wennalle Messungen durchgeführt wurden, öffnet das Gerät
automatisch ein neues Los mit einem Link zum ursprünglichen
Losnamen. Beispiel: das Neue Los_1 heißt "Neues Los_2","Neues Los_3" usw.
Speichern Sie jeden Messwert einzeln oder speichern Sie den Durchschnitt
einer vorgegebenen Anzahl von Messwerten mit Hilfe der Funktion "Gezählter
Durchschnitt".

12 DIAGRAMMANZEIGE
12.1 BALKENGRAFIKDie Balkengrafik zeigt eine analoge
Darstellung des Dickenwertes sowie denhöchsten, niedrigsten unddurchschnittlichen Messwert an. Zur
Anzeige der Balkengrafik:
1

Drücken Sie auf die Taste Display undwählen Sie “Messwerte & Balkengrafik”

Wenn ein Messwert außerhalb der eingestellten Grenzwerte liegt, werden der
weiße Balken und der Messwert rot angezeigt.
Im "Scan-Modus" (Modell T) wird die Balkengrafik automatisch während jedes
Scans angezeigt.

12.2 DIAGRAMM (MODELL S & T)
Anzeigen des Diagramms derletzten 20 Messwerte:123Drücken Sie die Taste "LOS".
Wählen Sie "Neues Los" oder “Öffne
Vorhandenes Los".
Drücken Sie auf die Taste Display undwählen Sie “Messwerte & Diagramm”.
Rote Punkte verweisen (sofern festgelegt) auf außerhalb der Grenzwerte desLoses liegende Messwerte.

www.elcometer.com

de-8

R

12 DIAGRAMMANZEIGE (Fortsetzung)12.3 LOS-GRAFIK (MODELL T)
Anzeigen der Los-Grafik:
1
2



de-9

Wählen Sie den entsprechenden LosNamen unter Los/Los Betrachten.
Wählen Sie "Los-Grafik".
Rote Balken verweisen (sofern festgelegt) auf
außerhalb der Grenzwerte des Loses liegende
Messwerte.
Drücken Sie zum Betrachten einzelner
Messwerte auf die Taste Zoom+ und dann auf
 oder .

www.elcometer.com

www.elcometer.com

Statistik Auswählen
Alle PSPC Werte
Anzahl Messwerte
Mittelwert
Kleinster Messwert
Grösster Messwert
StandardabweichungVariationskoeffizient
Unterer Grenzwert
Anzahl < Grenzwert
Oberer Grenzwert
Anzahl > Grenzwert
EIV
NDFT Wert%>=NDFT% 90-100% NDFTIMO 90:10 OK/Nicht OK

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

wo Nr. = 1, 2 oder 3

Kal Speicher nGespeichert h:min T/M/J
WählenKalibrieren
Kal-MethodeKalibrierung Testen
Name Neu Kal Speicher n
Kalibrierdaten Betrachten
Löschen Kal Speicher n

Kal-Methode
Null
Glatt
Rau/2-Punkt
Null-Offset
Auto

Statistik
Statistik Löschen
Zeige Einzelne Werte
Zeige Alle Werte
Statistik Auswählen
Gezählt. Durchschnitt

Kalibrierung
Kalibrieren
Kal-MethodeKal Speicher 1
Kal Speicher 2
Kal Speicher 3
Fabrik-Kalibrierung
Kalibriersperre
Kalibrierung Testen
Im-Kal-Bereich Prüfung

Los
Freier Speicher
Neues LosÖffne vorhandenes Los
Los-Modus Verlassen?
Bearbeite Los
Kopiere Los
Betrachte Los
Gelöschter Messwert

Display
Statistik Löschen
Nur Messwerte
Messwerte & def. Statistik
Messwerte & Diagramm
Messwerte & Balkengrafik
Los-Grafik
Statistik

Menü
Letzten Wert Löschen
Einstellung
Grenzwerte Einstellen
Bluetooth
Infos
Reset
Notleuchte

○

○

○
○
○
○

○

Gelöschter Messwert
Löschen Mit Anzeige○
Löschen Ohne Anzeige○

Bearbeite Los
Los Umbenennen
Lösche Losinhalt
Los Löschen
Betrachte Los
Los 1
Statistik
Los-Information
Kalibrier-Information
Messwerte
Los-Grafik

Neues LosÖffnen Los 1
Neubenennung Los 1
Los-Kalibrierung
Los-Typ
Los-Grenzwerte
Feste Losgröße
NAVSEA

Geräte-Infos
Sonden-Infos
Kontakt
Legal

Infos

BluetoothBluetooth Aktivieren○
Bluetooth ID
xxxxxxBluetooth CodexxxxxxAI

Grenzwerte Einstellen
Grenzwerte: An○
Obere Setzen
Untere Setzen
NDFT Einstellen

Einheiten
SpracheZeit und Datum
Piepserlautstärke
Bildschirmeinstellungen
Sonde
Mess-Modus
Automatische Abschaltung ○Kalibriersperre
NAVSEA○

Einstellung○

Zeit und Datum
Zeit Einstellen
Datum Einstellen
Format Einstellen
Zeit und Datum Zeigen

Los-Grenzwerte
Grenzwerte: An
Obere Setzen
Untere Setzen
NDFT Einstellen

Los-Typ
NormalGezählt. Durchschnitt
IMO PSPC

Los-Kalibrierung
Kalibrieren
Kal-MethodeNutzen Kal Speicher 1
Nutzen Kal Speicher 2
Nutzen Kal Speicher 3

Legal
PatenteGesetzliche Hinweise
Behördlich

○

○
○
○

○
○
○

○
○
○
○

Mess-Modus
Standard-Modus
Wiederholautomatik
Scan-Modus
Scanmodus mit Haltefkt
Kalibriersperre
PIN aktivieren

○
○

○
○
○
○

Ferritisch F1
Ferritisch F2

Sonde

Sonde
Auto FNF
Ferritisch F
Nicht-Ferritisch N
Sondeneinschaltung an

Bildschirmeinstellungen
Bildschirmhelligkeit
Abschaltzeit Bildschirm
Auto-Bildschirmdrehung○
Eingangsbild
○

○
○
○
○

Einheiten
µm
mm
mils
inch

Sonde

Behördlich

Kal-Methode
Null
Glatt
Rau/2-Punkt
Null-Offset
Auto
ISO
SSPC PA2
SchwedischAustralisch

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Die Menüanzeige hängt ab von derangeschlossenen Sonde.

R

13 MENÜSTRUKTUR - ELCOMETER 456 MODELL T

de-10

de-11

Statistik Auswählen
Alle PSPC Werte
Anzahl Messwerte
Mittelwert
Kleinster Messwert
Grösster Messwert
StandardabweichungVariationskoeffizient
Unterer Grenzwert
Anzahl < Grenzwert
Oberer Grenzwert
Anzahl > Grenzwert
EIV
NDFT Wert%>=NDFT% 90-100% NDFTIMO 90:10 OK/Nicht OK

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

Kal-Methode
Null
Glatt
Rau/2-Punkt
Null-Offset
Auto

Statistik
Statistik Löschen
Zeige Einzelne Werte
Zeige Alle Werte
Statistik Auswählen
Gezählt. Durchschnitt

○
○
○
○
○
Kalibrierung
Kalibrieren
Kal-MethodeFabrik-Kalibrierung
Kalibriersperre
Kalibrierung Testen

Los
Freier Speicher
Neues LosÖffne vorhandenes Los
Los-Modus Verlassen?
Bearbeite Los
Betrachte Los
Gelöschter Messwert

Display
Statistik Löschen
Nur Messwerte
Messwerte & def. Statistik
Messwerte & Diagramm
Messwerte & Balkengrafik
Statistik

Menü
Letzten Wert Löschen
Einstellung
Grenzwerte Einstellen
Bluetooth
Infos
Reset
Notleuchte

○

○
○
○
○

Gelöschter Messwert
Löschen Mit Anzeige○
Löschen Ohne Anzeige○

Statistik
Los-Information
Kalibrier-Information
Messwerte

Betrachte Los
Los 1

Bearbeite Los
Lösche Losinhalt
Los Löschen

Neues LosÖffnen Los
Los-Kalibrierung
Los-Typ

Geräte-Infos
Sonden-Infos
Kontakt
Legal

Los-Typ
NormalGezählt. Durchschnitt
IMO PSPC

Los-Kalibrierung
Kalibrieren
Kal-MethodePatenteGesetzliche Hinweise
Behördlich

Legal

Kalibriersperre
PIN aktivieren

Sonde

Sonde
Auto FNF
Ferritisch F
Nicht-Ferritisch N
Sondeneinschaltung an

BluetoothBluetooth Aktivieren○
Bluetooth ID
xxxxxxBluetooth CodexxxxxxAI
Ferritisch F1
Ferritisch F2

Bildschirmeinstellungen
Bildschirmhelligkeit
Abschaltzeit Bildschirm
Auto-Bildschirmdrehung○
Eingangsbild○

Grenzwerte Einstellen
Grenzwerte: An○
Obere Setzen
Untere Setzen
NDFT Einstellen

Infos

○

Zeit und Datum
Zeit Einstellen
Datum Einstellen
Format Einstellen
Zeit und Datum Zeigen

Einheiten
SpracheZeit und Datum
Piepserlautstärke
Bildschirmeinstellungen
Sonde
Automatische Abschaltung ○Kalibriersperre

○
○
○

○
○

○
○
○
○

○
○
○
○

Einstellung

Einheiten
µm
mm
mils
inch

Sonde

Behördlich

Kal-Methode
Null
Glatt
Rau/2-Punkt
Null-Offset
Auto
ISO
SSPC PA2
SchwedischAustralisch

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Die Menüanzeige hängt ab von derangeschlossenen Sonde.

R

14 MENÜSTRUKTUR - ELCOMETER 456 MODELL S

www.elcometer.com

www.elcometer.com

Statistik Auswählen
Anzahl Messwerte
Mittelwert
Kleinster Messwert
Grösster Messwert
StandardabweichungVariationskoeffizient
EIV

○
○
○
○
○
○
○

Kal-Methode
Null
Glatt
Rau/2-Punkt

Statistik
Statistik Löschen
Zeige Einzelne Werte
Zeige Alle Werte
Statistik Auswählen
Gezählt. Durchschnitt○○○

LöscheKalibrierung
Kalibrieren
Kal-MethodeFabrik-Kalibrierung
Kalibriersperre
Kalibrierung Testen

Abbr.

51.5µmDaten
Letzte Fünf Messwerte
50.5 µm52.3 µm51.0 µm51.8 µm

Display
Statistik Löschen
Nur Messwerte
Messwerte & def. Statistik
Messwerte & Diagramm
Statistik

Menü
Letzten Wert Löschen
Einstellung
Infos
Reset
Notleuchte

○

OK

○
○
○
Geräte-Infos
Sonden-Infos
Kontakt
Legal

Infos

Einheiten
SprachePiepserlautstärke
Bildschirmeinstellungen
Sonde
Automatische Abschaltung ○Kalibriersperre

Einstellung

Einheiten

○
○
○
○

Sonde

PatenteGesetzliche Hinweise
Behördlich

Legal

Kalibriersperre
PIN aktivieren

Ferritisch F1
Ferritisch F2

Sonde
Auto FNF
Ferritisch F
Nicht-Ferritisch N
Sondeneinschaltung an

○
○

○
○
○
○

Bildschirmeinstellungen
Bildschirmhelligkeit
Abschaltzeit Bildschirm
Auto-Bildschirmdrehung○
Eingangsbild
○

µm
mm
mils
inch

Sonde

Behördlichxxxxx

Die Menüanzeige hängt ab von derangeschlossenen Sonde.

R

15 MENÜSTRUKTUR - ELCOMETER 456 MODELL B

de-12

R

16 DATENDOWNLOAD UND MESSGERÄTUPGRADE
16.1 MIT ELCOMASTER®
Mit Hilfe der ElcoMaster® - im Lieferumfang beim Elcometer 456
Modell S & T, und erhältlich als kostenloser Download unter
www.elcometer.com - können alle Geräte Daten zur Archivierung undzur Erstellung von Berichten auf einen PC übertragen. Die Daten
können über eine USB-Schnittstelle oder über eine Bluetooth®
(S & T Modell) übertragen werden. Weitere Informationen zu
ElcoMaster® finden Sie unter www.elcometer.com
16.2 MIT ELCOMASTER® MOBILE APPS (MODELL S & T)
ElcoMaster® Mobile Apps für Android™ oder iOS sind die ideale
Lösung beim Arbeiten auf der Baustelle oder vor Ort und bieten
folgende Funktionen:






Direktes Speichern von LiveMesswerten auf einem Mobilgerät undAblage in Losen zusammen mit GPSKoordinaten.
Einfügen von Fotos der Prüffläche.
Zuordnen von Messwerten zu einer
Karte, einem Foto oder Diagramm.
Inspektionsdaten können zur weiteren
Auswertung und zum Erstellen vonBerichten vom Mobilgerät auf einen
PC übertragen werden.

Weitere Information zu ElcoMaster® Mobile
Apps finden Sie auf www.elcometer.com.
Geeignet für Smartphones undTablets, die Android 2.1 oderhöher verwenden. Laden Sie die
App zum Installieren überwww.elcometer.com oder Google
Play™ Store herunter und folgen
Sie den Anleitungen auf dem
Bildschirm.

de-13

www.elcometer.com

R

16 DATENDOWNLOAD UND MESSGERÄTUPGRADE (Fort.)
Geeignet für iPhone 6 Plus,
iPhone 6, iPhone 5s, iPhone 5c,
iPhone 5, iPhone 4s, iPhone 4,
iPad Air 2, iPad mini 3, iPad Air,
iPad mini 2, iPad (3. und 4.
Generation), iPad mini, iPad 2,
und iPod touch (4. und 5.
Generation). Sie die App überden www.elcometer.com oderden App Store herunter undfolgen Sie den Anleitungen auf
dem Bildschirm.
16.3 UPGRADE IHRES MESSGERÄTS
Die Messgerät-Firmware kann mit ElcoMaster® auf die jeweils neueste
Version aktualisiert werden, sobald sie verfügbar wird. Wenn das
Messgerät mit einem PC mit einer Internet-Verbindung verbunden
wird, informiert Sie ElcoMaster® über etwaige verfügbare Updates.
17 TECHNISCHE DATEN
Batterietyp

2 x AA Batterien, auch wiederaufladbare Batterienkönnen eingesetzt werden.

Betriebstemperatur-10 bis 50°C (14 bis 122°F)

Relative
Luftfeuchte

0 bis 95%

Geräteabmessungen14,1 x 7,30 x 3,70cm(5,55 x 2,87 x 1,46")

Gerätegewicht

Integrierte Sonde: 156g (5,5oz)
Separate Sonde: 161g (5,68oz)(mit gelieferten
Batterien)

Kann in Übereinstimmung mit folgenden Standards eingesetzt werden:
AS 2331.1.4, AS 3894.3-B, AS/NZS 1580.108.1, ASTM B 499, ASTM D 1186-B,
ASTM D 1400, ASTM D 7091, ASTM E 376, ASTM G 12, BS 3900-C5-6B,
BS 3900-C5-6A, BS 5411-11, BS 5411-3, BS 5599, DIN 50981, DIN 50984,
ECCA T1, EN 13523-1, IMO MSC.215(82), IMO MSC.244 (83), ISO 1461,
ISO 19840, ISO 2063, ISO 2178, ISO 2360, ISO 2808-6A, ISO 2808-6B,
ISO 2808-7C, ISO 2808-7D, ISO 2808-12, NF T30-124, SS 184159, SSPC PA 2,
US Navy PPI 63101-000, US Navy NSI 009-32

www.elcometer.com

de-14

R

18 RECHTLICHE HINWEISE UND BEHÖRDLICHE INFORMATIONEN
Konformitätserklärung:
Das Elcometer 456 Modell B entspricht den Anforderungen folgender EU-Anordnungen:2014/30/EU
Elektromagnetische Verträglichkeit
2011/65/EU
Beschränkung der Verwendung bestimmter gefährlicher Stoffe
Die Elcometer 456 Modelle S & T entsprechen den Anforderungen der folgenden EU Anordnungen:2014/53/EU
Funkanlagen2011/65/EU
Beschränkung der Verwendung bestimmter gefährlicher Stoffe
Konformitätserklärungen können mit folgenden Links heruntergeladen werden:
Modell B: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/German/DoC_456C_B.pdf
Modell S und T: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/German/DoC_456C_S&T.pdf
Betriebsfrequenzbereich: 2,402 - 2,480 MHz
Maximale Sendeleistung: <4 dBm
Dieses Produkt ist ein ISM-Gerät der Klasse B, Gruppe 1 gemäß CISPR 11.
Produkt der Klasse B: Es ist für den Gebrauch in Wohnbereichen und in Bereichen geeignet, die direkt mit einem
Niederspannungs-Stromversorgungsnetz verbunden sind, das Gebäude für den häuslichen Gebrauch versorgt.
ISM-Produkt der Gruppe 1: Ein Produkt, in dem beabsichtigt konduktiv gekoppelte Funkfrequenzenergie erzeugt und/oder
verwendet wird, die für die interne Funktion der Ausrüstung selbst erforderlich ist.
Der USB-Anschluss dient nur zur Datenübertragung und darf nicht über einen USB-Netzadapter am Netzstromangeschlossen werden.
Das Gerät ist kompatibel zum Teil 15 der FCC Richtlinien. Der Betrieb setzt die folgenden Bedingungen voraus: (1) es darf
keine schädliche Störung vom Gerät ausgehen und (2) das Gerät muss jegliche Störung tolerieren, eingeschlossen Störungen
die einen nicht erwünschten Betrieb verursachen.
Der Zugriff auf das ACMA-Konformitätszeichen erfolgt über: Menü/Infos/Legal/Behördlich
Elcometer 456 Modell S und T: Das Giteki-Zeichen, seine Verordnungsnummer, die FCC-ID und Bluetooth SIG QDID sindabrufbar über: Menü/Infos/Rechtliches/Prüfzeichen
Hinweis: Dieses Gerät wurde erfolgreich getestet auf die Erfüllung der Beschränkungen eines digitalen Gerätes der Klasse B
gemäß Teil 15 der FCC Richtlinien. Diese Beschränkungen dienen dem angemessenen Schutz gegen schädliche Interferenz
in einer stationären Installation. Dieses Gerät erzeugt, nutzt und strahlt möglicherweise Hochfrequenzenergie und kann bei
bei nicht vorschriftsgemäßer Installation oder Nutzung schädliche Störungen des Funkverkehrs verursachen. Es gibt jedochkeine Garantie, dass bei bestimmten Installationsarten keine Interferenz auftritt. Falls das Gerät schädliche Störungen desRadio- oder Fernsehempfangs verursacht, was man feststellen kann durch Ein- und Ausschalten des Gerätes, sollte derNutzer die Störung durch eine oder mehrere der folgenden Maßnahmen beseitigen:– Die Empfangsantenne neu einstellen oder neu positionieren.
– Das Gerät und das Empfangsgerät weiter auseinander stellen.
– Das Gerät mit einem anderen Stromkreis verbinden als den den das Empfangsgerät nutzt.
– Den Händler oder einen erfahrenen Radio-/Fernsehtechniker um Hilfe bitten.
Zur Erfüllung der FCC-Strahlenbelastungsrichtlinien für mobile und Basisstation-Sendegeräte sollte während des Betriebs ein
Abstand von mindestens 20 cm zwischen der Antenne dieses Geräts und Personen eingehalten werden. Zur Gewährleistung der
Konformität wird ein Betrieb mit einem geringeren als diesem Abstand nicht empfohlen. Die für diesen Sender verwendeten
Antennen dürfen nicht am selben Ort wie eine andere Antenne oder ein anderer Sender platziert oder in Verbindung mit diesen
betrieben werden.
Änderungen die nicht ausdrücklich von Elcometer Ltd. genehmigt sind könnten die Nutzungsberechtigung gemäß der FCC
Richtlinien zum Erlöschen bringen.
Dieses Gerät entspricht der/den lizenzfreien RSS-Norm/en von Industry Canada. Sein Betrieb ist vorbehaltlich der beiden
folgenden Bedingungen zulässig: (1) Dieses Gerät darf keine Interferenzen verursachen und (2) dieses Gerät muss jeglicheInterferenzen annehmen, einschließlich Interferenzen, die einen unerwünschten Betrieb des Geräts verursachen können.
Dieses digitale Gerät der Klasse B entspricht der Norm CAN ICES-3 (B)/NMB-3(B).
und ElcoMaster® sind eingetragene Markenzeichen der Elcometer Limited, Edge Lane, Manchester, M43 6BU,
Großbritannien und Nordirland.
ist eine Handelsmarke im Eigentum der Bluetooth SIG Inc und lizenziert für Elcometer Limited.
Elcometer 456 Modell S und T: Geeignet für iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPhone 4s, iPhone 4, iPad
Air 2, iPad mini 3, iPad Air, iPad mini 2, iPad (3. und 4. Generation), iPad mini, iPad 2, und iPod touch (4. und 5. Generation).
“Made for iPod”, ”Made for iPhone” und “Made for iPad” bedeutet, dass elektronisches Zubehör speziell für die Verbindung mit
iPod, iPhone respektive iPad konzipiert und vom Entwickler als die Apple-Leistungsstandards erfüllend zertifiziert wurde.
Apple ist nicht für den Betrieb dieses Geräts oder seine Konformität mit Sicherheits- und aufsichtsbehördlichen Standards
verantwortlich. Beachten Sie bitte, dass sich der Gebrauch dieses Zubehörs in Verbindung mit iPod, iPhone oder iPad auf die
Wireless-Leistung auswirken könnte.
iPad, iPhone und iPod touch sind in den USA und anderen Ländern eingetragene Markenzeichen der Apple Inc.
App Store ist ein den USA und anderen Ländern eingetragenes Markenzeichen der Apple Inc.
Google Play ist ein Markenzeichen der Google Inc.
Alle anderen Handelsmarken sind anerkannt.
R

Das Elcometer 456 ist in einem Karton verpackt. Stellen Sie bitte sicher, dass alle Verpackung auf umweltverträgliche Weise
entsorgt wird. Lassen Sie sich von Ihrer örtlichen Umweltbehörde weiterberaten.
Hauptniederlassung: Elcometer Limited, Edge Lane, Manchester, M43 6BU, Großbritannien und Nordirland.

TMA-0488-02 Ausgabe 17 - Text mit Umschlag Nr. 22373-02

R

Instrucciones de uso
Medidor de espesor de revestimientos

Elcometer 456

www.elcometer.com

R

CONTENIDO
Sección

Página

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18

Presentación general del medidor
Contenido de la caja
Utilizar el medidor
InicioConectar la sonda
Tomar una lectura
Calibrar el medidor
Bloquear y desbloquear calibración
Métodos de calibración
Modos de Medición
Lotes
Desplegar Gráficas
Estructura del Menú - Modelo T
Estructura del Menú - Modelo S
Estructura del Menú - Modelo B
Descarga de datos y actualización del medidor
Especificaciones técnicas
Avisos legales e información sobre la normativa

es-2
es-2
es-3
es-3
es-4
es-4
es-5
es-5
es-6
es-7
es-8
es-8
es-10
es-11
es-12
es-13
es-14
es-15

Pb

Hg

Para evitar cualquier duda, por favor refiérense a la versión original en inglés.
El Elcometer 456 está disponible en 3 modelos. Este manual de instrucciones ha sido realizado para el modelo T.
Los modelos B y S están referenciados donde es adecuado.
Dimensiones del Medidor: 141 x 73 x 37mm (5.55 x 2.87 x 1.46")
Peso del Medidor: Integral: 156g (5.5oz) con pilas incluidas; Separado: 161g (5.68oz) con pilas incluidas.
Patentes aplicables:
US6243 661; US5886522; US6762603; US7606671; GB2306009; GB2367135;
GB2342450, DE10131827© Elcometer Limited 2012 - 2018. Todos los derechos reservados. Ninguna parte de este documento podrá
reproducirse, difundirse ni almacenarse (en un sistema de recuperación u otro) ni traducirse a otro idioma, deninguna forma ni medio (electrónico, mecánico, magnético, óptico, manual u otro), sin el previo consentimiento
por escrito de Elcometer Limited.

es-1

www.elcometer.com

R

1 PRESENTACIÓN GENERAL DEL MEDIDOR
6

1
2

7
3
8

4

5

1
2
3
4
5
6
7
8

Indicadores LED - Rojo (izquierda), Verde (derecha)
Pantalla Color
Teclas multifunciones
Botón On/Off
Conexión para Sonda integrada / separada
Toma de transferencia de datos USB (debajo de la tapa)
Compartimiento de pilas (¼ de vuelta para apertura / cierre)
Enganche para correa de muñeca

2 CONTENIDO DE LA CAJA











Medidor de espesor de revestimientos Elcometer 456
Galgas de calibración (Medidores con sonda integrada)
Correa de muñeca
Estuche de protección (modelos B, S y T)
Maletín de transporte (modelo T)
1 x Protector de pantalla (modelos S y T)
2 x pilas AA
Cable USB y Programa ElcoMaster® (S y T)
Certificado de prueba
Instrucciones de uso

www.elcometer.com

es-2

R

3 UTILIZAR EL MEDIDORa Indicador de nivel de pilab Bluetooth activado - Gris: noemparejado, Naranja: emparejado
Tipo de substrato - F, N, FNF

cd Método de Calibracióne Alerta Medición Fuera deCalibración Activada
f Limite máximo activado
g Unidad de medición -Modelo
BST
o
ST

a
b
c

n

BST
BST
T
ST
BST

m
l
k

d
e
f
g
h
i
j

µm, mils, mm, inch

h Tipo de lote - normal, media, IMO ST
i Tecla MenúBST
j Tecla Pantalla
BST
k Tecla Lotes / Datos
BSTl Tecla de calibración
BSTm Valor de la lectura
BSTn Estadísticas seleccionables por BST
el usuario - 4 lineaso Nombre de lote
T

p

q

r

(utilizando la función lotes)pFecha y HoraST

(fuera de la función lotes)
Grafico Barra - Lectura mas alta,mas baja y promedio

BST

q Limite mínimo activador Modo de Medición - Estándar,

ST
T

s

t

Auto Repetición, Escan
Graf Ejecución - ultimas 20 lecturas ST

st Limites máximo y mínimo
activadosST

4 INICIO
4.1 ASEGURANDO QUE SU MEDIDOR DISPONGA DEL ULTIMOFIRMWARE Y ACTUALIZANDO SU MEDIDOR
Para asegurar que su medidor disponga del firmware más
actualizado, permitiéndole beneficiarse de las ultimas características yfuncionalidad, le recomendamos conectar regularmente el equipo a
ElcoMaster® y antes de su primer uso.
es-3

www.elcometer.com

R

4 INICIO (continuación)
Simplemente conecte el medidor vía USB a un ordenador conectado ainternet y ejecutando ElcoMaster® utilizando la función ‘Conectar
medidor’. Si hay disponible una versión más actualizada del firmware
del medidor, la función ‘Actualizar medidor’ se mostrará a la derechade los detalles del medidor. Haga clic en ‘Actualizar medidor’ para
instalar el último firmware.
4.2 SELECCIÓN DEL IDIOMA
1
Presione y mantenga la tecla ON/OFF hasta que el logo
Elcometer aparezca.
2
Seleccione su idioma utilizando las teclas .
3Sigan las instrucciones de los menús en la pantalla.
Para acceder al menú idiomas cuando el medidor esté en un idioma
extranjero:1
Apagar el medidor.
2
Presione y sostenga la tecla izquierda y encienda el medidor.
3
Seleccione su idioma utilizando las teclas .
5 CONECTAR LA SONDA
MEDIDORES CON SONDA SEPARADA ÚNICAMENTE
1
Rotar enchufe de la sonda para alinear las clavijas.
2
Enroscar la sonda - en el sentido de las agujas de un reloj.

2

1

2

6 TOMAR UNA LECTURA
1
2
3

1

Sostenga la sonda por el mango.
Para tomar una lectura, coloque la sonda hacia abajo sobre lasuperficie manteniéndola perpendicular.
Para las siguientes lecturas, levante la punta de la sonda yvuelva a colocarla en la superficie con revestimiento.

www.elcometer.com

es-4

R

6 TOMAR UNA LECTURA
 ES ACONSEJADO
Sostener la sonda por el mango.
••
Colocar delicadamente la sonda sobre la superficie.
•
Dejar el mango tocar la superficie - para mejorar la precisión.
 EVITAR
Arrastrar la sonda en la superficie con revestimiento.
••
Golpear la superficie con fuerza con la sonda.
•
Sobrevolar la superficie con la sonda ya que esto podría
generar lecturas erróneas.


La pantalla se obscurece si el medidor esta inactivo durante más de 15
segundos y se vuelve "negra" si queda inactivo durante el período definido en
Menú/Ajustes/Ajustes de Pantalla/Tiempo Apagado Pantalla. Despierte elmedidor ya sea presionando cualquier tecla o con un pequeño golpe.
El medidor se apagará automáticamente después de 5 minutos de inactividad.
- - - indica lecturas fuera del rango de la sonda.

7 CALIBRAR EL MEDIDOR
1
Presionar la tecla Cal.
2
Para métodos de calibración alternativos,
seleccione Cal/Método Cal.
3
Seleccione Calibrar y siga las
instrucciones en la pantalla.
4
Cuando sea indicado coloque lasonda al centro de las galgas.



90°

Las galgas pueden superponerse.
No todos los métodos de calibración están disponibles para todos los tipos demedidores ver 9 Métodos de calibración.

8 BLOQUEAR Y DESBLOQUEAR CALIBRACIÓN
Para establecer el código PIN de bloqueode calibración:
1
Presione la tecla de Menú yseleccione Ajustes/Bloquear
Calibración.
2
Establecer el Código PIN de cuatro
dígitos usando las teclas  paraseleccionar del 0 al 9 y la tecla  para
moverse del primero al cuarto digito†.
3
Presione Ok, Escape para cancelar o Ajustar (adjust) paracorregir el código NIP.
†

La tecla  aparecerá cuanto la “ X “ cambié a un numero.

es-5

www.elcometer.com

R

8 BLOQUEAR Y DESBLOQUEAR CALIBRACIÓN (cont.)
Para desbloquear calibración:
1
Presione la tecla Cal y seleccione Bloquear calibración.
2
Ingrese los cuatro dígitos de su código PIN, si se ha establecido,
usando las teclas  para seleccionar del 0 al 9 y la tecla para mover del primero al cuarto dígito†.
3
Presione Ok o Escape para cancelar.
Para deshabilitar el código PIN bloqueo de calibración:
1
Presione la tecla de Menú y seleccione Ajustes/Bloquear
Calibración.
2
Ingrese los cuatro dígitos de su código PIN.


Si el usuario olvida o extravía el código PIN, este se puede deshabilitar vía
ElcoMaster®. Utilizando el cable USB, simplemente conecte el medidor a una PC
con ElcoMaster® versión 2.0.33 o mayor instalado y seleccione Editar/Eliminar
PIN de calibración.

9 MÉTODOS DE CALIBRACIÓN

†MÉTODO DE
CALIBRACIÓN

MODELODESCRIPCIÓNCero

B, S, T

Una calibración en un punto - ideal parasuperficies lisas, simplemente coloque lasonda en el substrato sin revestimiento y elmedidor se calibrará automáticamente

Lisa

B, S, T

Una calibración en dos puntos - el usuario
selecciona una galga de calibración adecuada
y un substrato liso sin revestimiento (cero)Rugosa/ 2 puntos

B, S, TEl método de calibración ideal para
substratos rugosos o perfilados, utilizando
dos galgas con espesores conocidos - una
por encima, y otra por debajo del espesor depelícula seca que desea obtenerCompensación
cero

S, T

Método de calibración para mediciones
cuando el perfil / rugosidad del sustrato sedesconoce, o no es accesible. Un valor decompensación definido por el usuario se
añade al valor de la medición de lectura

La tecla  aparecerá cuanto la “ X “ cambié a un numero.

www.elcometer.com

es-6

R

9 MÉTODOS DE CALIBRACIÓN (continuación)
MÉTODO DE
CALIBRACIÓN

MODELODESCRIPCIÓN

Auto

S, T

Ideal para inspecciones repetitivas. El usuario
almacena espesores conocidos de galgas decalibración en el medidor. Al calibrar, el
usuario tiene que calibrar el medidor y elmedidor ajusta automáticamente las lecturas
con los espesores de las galgas
almacenados - simplifica y acelera
considerablemente el proceso de calibración

ISO

S, T

Establece el método de calibración a
Compensación Cero y la Media calculada a
5 lecturas - de acuerdo con ISO19840

SSPC PA2

S, T

Establece el método de calibración aRugosa/ 2 puntos y la Media calculada a 3
lecturas - de acuerdo con SSPC PA2

Sueco

S, T

Establece el método de calibración a Rugosa/
2 puntos y la Media calculada a 5 lecturas - deacuerdo con los Estándares SS

Australiano

S, T

Establece el método de calibración a
Compensación Cero y la Media calculada a 5
lecturas - de acuerdo con los Estándares AS

10 MODOS DE MEDICIÓN (MODELO T)
10.1 MODOS DE MEDICIÓN
Tres modos de medición disponibles: “Modo Estándar”, “Modo Auto
Repetición” y “Modo Escan”.
Para seleccionar el modo de medición, presione Menú/Ajustes/Modode Medición.
“Modo Auto Repetición” y “Modo Escan” están disponibles solamente con lasonda Ultra/Escan Elcometer 456. Para más información contacte a Elcometero visite www.elcometer.com.

10.2 MODO NAVSEA
Muestra la lectura redondeada a la milésima (µm) más cercana en
valores por encima de 10 milésimas (254µm) y 0.1 por debajo de
10 milésimas (254µm).
Para seleccionar el modo NAVSEA, presione Menú/Ajustes/NAVSEA o
Lote/Nuevo Lote/NAVSEA.
www.elcometer.com
es-7

R

11 LOTES (MODELO S Y T)
1
Para utilizar la función memoria en lotes, pulsar la tecla Lotes.
2
Seleccione “Nuevo Lote” o “Abrir Lote Existente” para añadir
lecturas.
3
Copiar y revisar datos del lote.
4
Seleccione “Editar Lote” para cambio de nombre, borrar lecturasdel lote o borrar un lote.
5
El Tamaño de lote fijo permite a los usuarios pre-definir elnúmero de lecturas que se almacenan en un lote. Una vez que
todas las lecturas se han tomado, el medidor abreautomáticamente un nuevo lote con una referencia al nombre
original del lote. Por ejemplo el lote NewBatch_1 se convierte en
NewBatch_2, NewBatch_3, etc.


Guarde cada lecturas en la memoria o almacene la media de un número delecturas predefinido con la función Promedio contado.

12 DESPLEGAR GRÁFICAS
12.1 GRAFICO BARRAEl Grafico Barra se despliega en una
representación analógica del valor deespesor junto con la lectura mas alta, masbaja y promedio al ser tomadas la
mediciones. Para mostrar el Grafico Barra:
1

Presione la tecla de Pantalla yseleccione "Lecturas y Grafico Barra”.



Si una lectura esta fuera de los limites establecidos, la barra blanca y el valor dela lectura se vuelve de color rojo.
Cuando esta en modo de Escan (Modelo T), el grafico barra es desplegado
automáticamente durante cada escaneo.



12.2 GRAF EJECUCIÓN (MODELO S Y T)
Para mostrar la graf ejecución de las
20 ultimas lecturas:
12
3


Presionar la tecla Lote.
Seleccione “Nuevo Lote” o “Abrir
Lote Existente”.
Presione la tecla de Pantalla yseleccione "Lecturas Graf Ejecución”.
Puntos rojos significa que una lectura esta fuera de los límites del lote (si son
establecidos).

www.elcometer.com

es-8

R

12 DESPLEGAR GRÁFICAS (continuación)
12.3 GRAFICO DE LOTE (MODELO T)
Para desplegar la gráfica de los lotes delecturas:
1
2



es-9

Seleccione el nombre del loteapropiado de Lote/Revisar Lote.
Seleccione “Grafico de Lote”.
Columnas rojas significa que una lectura esta fuera de los límites del lote (si son
establecidos).
Presione la tecla de Zoom+ (maximizar) seguida por  o  para revisar
lecturas individuales como se requiera.

www.elcometer.com

www.elcometer.com

Seleccionar Estadísticas
Todos Valores PSPC○No De Lecturas
○
Media
○
Menor Lectura○
Mayor Lectura○
Desviación Estándar
○Coef. De Variación○
Valor Límite Mínimo
○No. Debajo Del Límite
○Valor Límite Máximo
○No. Encima Del Límite
○EIV○
Valor NDFT○
% Superior o = a NDFT○
De 90 a 100% de NDFT○IMO 90:10 ✓/✕
○

○
○
○
○
○

donde n = 1, 2, o 3

Memoria Cal n
Guardado hh:mm dd/mm/yy
Selecc.
Calibrar
Método Cal
Prueba de Calibración
Renombrar Memoria Cal n
Ver Datos Calibración
Borrar Memoria Cal n

Cero
Lisa
Rugosa / 2Puntos
Comp. Cero
Auto

Método Cal

Estadísticas
Borrar Estadísticas
Ver Seleccionadas
Ver Todas
Seleccionar Estadísticas
Promedio Contado

Calibración
Calibrar
Método Cal
Memoria Cal 1
Memoria Cal 2
Memoria Cal 3
Calibración De Fábrica
Bloquear Calibración
Prueba de Calibración
Verificar Calibración

Lote
Memoria Disponible
Nuevo Lote
Abrir Lote Existente¿ Salir De Lotes ?
Editar Lote
Copiar Lote
Revisar Lote
Lectura Borrada

Pantalla
Borrar Estadísticas
Solo Lecturas
Lecturas y Estad Selecc
Lecturas Graf Ejecución
Lecturas y Grafico Barra
Grafico de Lote
Estadísticas

Menú
Borrar Última Lectura
AjustesEstablecer Límites
Bluetooth
Información
ReajustarLuz De Emergencia

○

○

○
○
○
○

○

○

○

Lectura Borrada
Borrar Con Marcar
Borrar Sin Marcar

Revisar Lote
Lote 1
Estadísticas
Información Lote
Información Calibración
Mediciones
Grafico de Lote

Editar Lote
Renombrar Lote
Borrar Lote
Borrar Lote

Nuevo Lote
Abrir Lote 1
Renombrar Lote 1
Calibración de Lote
Tipo De Lote
Limites Del Lote
Tamaño Fijo De Lote
NAVSEA

Información
Información Medidor
Información Sonda
Contactos
Legalidades○○○

Bluetooth
Activar Bluetooth○
ID Bluetooth
xxxxxxCódigo Bluetooth
xxxxxxAI

UnidadesIdiomaHora y Fecha
Volumen De Pitido
Ajustes de Pantalla
Sonda
Modo de Medición
Apagado Automático
Bloquear Calibración
NAVSEA
Establecer Límites
Activar Límites
Elegir Máximo
Elegir Mínimo
Selecc. NDFT

Ajustes

○

○
○
○
○
○
○

Hora y Fecha
Ajustar Hora
Ajustar Fecha
Selecc. Formato
Mostrar Hora y Fecha
Ajustes de Pantalla
Luminosidad Pantalla
Tiempo Apagado Pantalla
Rotación Auto Pantalla
Abriendo Pantalla
Sonda
Auto FNF
Ferroso F
No Ferroso N
Encender con Sonda

○
○
○
○

Modo de Medición
Modo Estándar
Modo Auto Repetición
Modo Escan
Modo Escan Retención

Limites Del Lote
Activar Límites
Elegir Máximo
Elegir Mínimo
Selecc. NDFT

Tipo De Lote
NormalPromedio Contado
IMO PSPC○○○○

Calibración de Lote
Calibrar
Método Cal○
Usar Memoria Cal 1
Usar Memoria Cal 2○○
Usar Memoria Cal 3

Legalidades
Patentes
Notas Legales
Regulatorio

Bloquear Calibración
Activar código PIN○○

Ferroso F1
Ferroso F2

Sonda

○
○
○
○

Unidades
µm
mm
mils
inch

Sonda

Método Cal
Cero
Lisa
Rugosa / 2Puntos
Comp. Cero
Auto
ISO
SSPC PA2
Sueco
Australiano

QDID

Regulatorio

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

El menú que aparece en la pantalla
depende de la sonda conectada

R

13 ESTRUCTURA DEL MENÚ - ELCOMETER 456 MODELO T

es-10

es-11

Seleccionar Estadísticas
Todos Valores PSPC○No De Lecturas
○
Media
○
Menor Lectura○
Mayor Lectura○
Desviación Estándar
○Coef. De Variación○
Valor Límite Mínimo
○No. Debajo Del Límite
○Valor Límite Máximo
○No. Encima Del Límite
○EIV○
Valor NDFT○
% Superior o = a NDFT○
De 90 a 100% de NDFT○○IMO 90:10 ✓/✕Método Cal
Cero
Lisa
Rugosa / 2Puntos
Comp. Cero
Auto

Estadísticas
Borrar Estadísticas
Ver Seleccionadas
Ver Todas
Seleccionar Estadísticas
Promedio Contado

○
○
○
○
○
Calibrar
Método Cal
Calibración De Fábrica
Bloquear Calibración
Prueba de Calibración

Calibración

Lote
Memoria Disponible
Nuevo Lote
Abrir Lote Existente¿ Salir De Lotes ?
Editar Lote
Revisar Lote
Lectura Borrada

Pantalla
Borrar Estadísticas
Solo Lecturas
Lecturas y Estad Selecc
Lecturas Graf Ejecución
Lecturas y Grafico Barra
Estadísticas

Menú
Borrar Última Lectura
AjustesEstablecer Límites
Bluetooth
Información
ReajustarLuz De Emergencia

○

○
○
○
○

○

○

Lectura Borrada
Borrar Con Marcar
Borrar Sin Marcar

Revisar Lote
Lote 1
Estadísticas
Información Lote
Información Calibración
Mediciones

Editar Lote
Borrar Lote
Borrar Lote

Nuevo Lote
Abrir Lote
Calibración de Lote
Tipo De Lote

Información
Información Medidor
Información Sonda
Contactos
Legalidades○○

Bluetooth
Activar Bluetooth○
ID Bluetooth
xxxxxxCódigo Bluetooth
xxxxxxAI

Establecer Límites
Activar Límites
Elegir Máximo
Elegir Mínimo
Selecc. NDFT

UnidadesIdiomaHora y Fecha
Volumen De Pitido
Ajustes de Pantalla
Sonda
Apagado Automático
Bloquear Calibración

Ajustes

Sonda

Tipo De Lote
NormalPromedio Contado
IMO PSPC

Calibración de Lote
Calibrar
Método Cal

Patentes
Notas Legales
Regulatorio

Legalidades

Bloquear Calibración
Activar código PIN

Ferroso F1
Ferroso F2

Sonda
Auto FNF
Ferroso F
No Ferroso N
Encender con Sonda

○
○
○

○
○

○
○
○
○

○
○

○

Hora y Fecha
Ajustar Hora
Ajustar Fecha
Selecc. Formato
Mostrar Hora y Fecha
Ajustes de Pantalla
Luminosidad Pantalla
Tiempo Apagado Pantalla
Rotación Auto Pantalla
Abriendo Pantalla

○
○
○
○

Unidades
µm
mm
mils
inch

Sonda

Método Cal
Cero
Lisa
Rugosa / 2Puntos
Comp. Cero
Auto
ISO
SSPC PA2
Sueco
Australiano

QDID

Regulatorio

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

El menú que aparece en la pantalla
depende de la sonda conectada

R

14 ESTRUCTURA DEL MENÚ - ELCOMETER 456 MODELO S

www.elcometer.com

www.elcometer.com

Seleccionar Estadísticas
No De Lecturas
○
Media
○
Menor Lectura○
Mayor Lectura○
Desviación Estándar
○Coef. De Variación
○
EIV
○

Cero
Lisa
Rugosa / 2Puntos

Método Cal

Estadísticas
Borrar Estadísticas
Ver Seleccionadas
Ver Todas
Seleccionar Estadísticas
Promedio Contado○○○

Calibración

Borrar

51.5µmCalibrar
Método Cal
Calibración De Fábrica
Bloquear Calibración
Prueba de Calibración

Esc

Datos
Últimas Cinco Lecturas
50.5 µm52.3 µm51.0 µm51.8 µm

Pantalla
Borrar Estadísticas
Solo Lecturas
Lecturas y Estad Selecc
Lecturas y Grafico Barra
Estadísticas

Menú
Borrar Última Lectura
Ajustes
Información
ReajustarLuz De Emergencia

○

Ok

○
○
○

Información
Información Medidor
Información Sonda
Contactos
Legalidades

Ajustes
UnidadesIdiomaVolumen De Pitido
Ajustes de Pantalla
Sonda
Apagado Automático
Bloquear Calibración○

Patentes
Notas Legales
Regulatorio

Legalidades

Bloquear Calibración
Activar código PIN

○
○

○
○
○
○

Sonda
Auto FNF
Ferroso F
No Ferroso N
Encender con Sonda
Sonda

○
○

Ajustes de Pantalla
Luminosidad Pantalla
Tiempo Apagado Pantalla
Rotación Auto Pantalla
Abriendo Pantalla

Ferroso F1
Ferroso F2

○
○
○
○

Unidades
µm
mm
mils
inch

Sonda

Regulatorio
xxxxx

El menú que aparece en la pantalla
depende de la sonda conectada

R

15 ESTRUCTURA DEL MENÚ - ELCOMETER 456 MODELO B

es-12

R

16 DESCARGA DE DATOS Y ACTUALIZACIÓN DEL MEDIDOR
16.1 CON ELCOMASTER®
Usar el ElcoMaster® - se suministra con el Elcometer 456
Modelo S y T, y está disponible para descarga gratis enelcometer.com - todos los medidores pueden transferir las lecturas a
un PC para archivarlas y generar reportes. Los datos pueden sertransferidos por USB o Bluetooth® (modelos S y T). Para obtener más
información sobre ElcoMaster®, visite www.elcometer.com
16.2 CON APLICACIONES MÓVILES ELCOMASTER®
(MODELOS S Y T)
Las aplicaciones móviles ElcoMaster® para Android™ o iOS, idóneas
para su uso en campo o en las instalaciones, permiten a los usuarios:






Almacenar lecturas dinámicas
directamente en un dispositivo móvil yguardarlas en lotes junto con
coordenadas de GPS.
Añadir fotografías de la superficie
sometida a prueba.
Representar lecturas en un mapa,
fotografía o diagrama.
Los datos de inspección puedentransferirse de móvil a PC pararealizar análisis adicionales y generar
informes.

Para obtener más información sobre lasaplicaciones móviles ElcoMaster®, visitewww.elcometer.com
Compatible con smartphones ytablets que ejecuten Android 2.1
o superior. Para instalarla,descárguela dewww.elcometer.com oempleando la aplicación Google
Play™ Store y siga las
instrucciones de la pantalla.
es-13

www.elcometer.com

R

16 DESCARGA DE DATOS Y ACTUALIZACIÓN DEL MEDIDOR (cont.)
Creado para iPhone 6 Plus,
iPhone 6, iPhone 5s, iPhone 5c,
iPhone 5, iPhone 4s, iPhone 4,
iPad Air 2, iPad mini 3, iPad Air,
iPad mini 2, iPad (3ª y 4ªgeneraciones), iPad mini,
iPad 2, y iPod touch (4ª y 5ªgeneraciones). Para instalarla,descárguela a través dewww.elcometer.com o de la App
Store y siga las instrucciones dela pantalla.
16.3 ACTUALIZACIÓN DEL MEDIDOREl usuario puede actualizar mediante ElcoMaster® el firmware a laversión más reciente cuando esté disponible. ElcoMaster® informará alusuario de la existencia de cualquier actualización cuando el medidor
se conecte a un PC con conexión a Internet.
17 ESPECIFICACIONES TÉCNICAS
Tipo de pila

2 x pilas AA, también se pueden usar pilas
recargables

Temperatura defuncionamientode -10 a 50°C (de 14 a 122°F)

Humedad relativa

de 0 a 95%

Dimensiones delMedidor14.1 x 7.30 x 3.70cm(5.55 x 2.87 x 1.46")

Peso del medidor

Integral: 156g (5.5oz)
Separado: 161g (5.68oz)(con pilas incluidas)

Puede ser utilizado de acuerdo con:
AS 2331.1.4, AS 3894.3-B, AS/NZS 1580.108.1, ASTM B 499, ASTM D 1186-B,
ASTM D 1400, ASTM D 7091, ASTM E 376, ASTM G 12, BS 3900-C5-6B,
BS 3900-C5-6A, BS 5411-11, BS 5411-3, BS 5599, DIN 50981, DIN 50984,
ECCA T1, EN 13523-1, IMO MSC.215(82), IMO MSC.244 (83), ISO 1461,
ISO 19840, ISO 2178, ISO 2063, ISO 2360, ISO 2808-6A, ISO 2808-6B,
ISO 2808-7C, ISO 2808-7D, ISO 2808-12, NF T30-124, SS 184159, SSPC PA 2,
US Navy PPI 63101-000, US Navy NSI 009-32

www.elcometer.com

es-14

R

18 AVISOS LEGALES E INFORMACIÓN SOBRE LA NORMATIVA
Declaración de conformidad:
Elcometer 456 Modelo B cumple los requisitos de las siguientes Directivas de la UE:
2014/30/EU
Compatibilidad Electromagnética
2011/65/EU
Restricciones a la utilización de determinadas sustancias peligrosas
Elcometer 456 Modelos S y T cumple los requisitos de las siguientes Directivas de la UE:
2014/53/EU
Equipos de radio
2011/65/EU
Restricciones a la utilización de determinadas sustancias peligrosas
Las Declaraciones de Conformidad pueden descargarse a través de:Modelo B: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/Spanish/DoC_456C_B.pdf
Modelos S y T: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/Spanish/DoC_456C_S&T.pdf
Banda de frecuencias de funcionamiento: 2,402 - 2,480 MHz
Potencia máxima transmitida: <4 dBm
Este producto es un equipo de Clase B, Grupo 1 ISM, conforme a las normas CISPR 11.
Producto de clase B: Es apto para su uso en entornos domésticos y establecimientos conectados directamente a una red desuministro de baja tensión que suministre a edificios dedicados a uso residencial.
Producto de Grupo 1 ISM: Producto que genera y/o utiliza intencionadamente energía de radiofrecuencia de acoplamiento
conductivo necesaria para el funcionamiento interno del propio equipo.
El USB es para transferencia de datos solamente y no debe conectarse a la red eléctrica mediante un adaptador de USB/redeléctrica.
Este equipo cumple con la Parte 15 de las Normas de la FCC. Su utilización está sujeta a las siguientes dos condiciones: (1)este equipo no debe causar interferencias perjudiciales y (2) este dispositivo debe aceptar cualquier interferencia recibida,incluyendo interferencias que puedan causar un funcionamiento no deseado.
Es posible acceder a la marca de cumplimiento ACMA mediante: Menú/Información/Legalidades/Regulatorio
Elcometer 456 Modelos S y T: Puede accederse a la marca Giteki, su número de reglamento, el ID de la FCC y el QDID deBluetooth SIG a través de: Menú/Acerca de/Legal/Regulaciones
NOTA: Este equipo ha sido probado y se encontró que cumple con los límites de un equipo digital de clase B conforme con elapartado 15 de las Normas de la FCC. Estos límites son diseñados para proporcionar una protección razonable contra lasinterferencias nocivas en una instalación residencial. Este equipo genera, usa y puede irradiar la energía de la frecuencia dela radio y si no ha sido instalado y usado de acuerdo con las instrucciones de uso, puede causar interferencias nocivas a las
comunicaciones por radio. Sin embargo, no hay garantía que la interferencia no ocurra en ninguna instalación. Si este equipocausa interferencias nocivas en la recepción de la radio o la televisión, lo que puede ser determinado apagando y
encendiendo el equipo, seria preferible que el usuario trate de corregir la interferencia intentando tomar una o varias de lasmedidas siguientes:– Reorientar o desplazar la antena de recepción.
– Aumentar la distancia entre el equipo y el receptor.
– Conectar el equipo en una toma de corriente en un circuito diferente al que el receptor esta conectado.
– Consultar con un vendedor o un Tecnico experimentado de televisiones y radios.
Para cumplir los requisitos de exposición a radiofrecuencias de la FCC para dispositivos móviles y de transmisión de estaciónbase, debe mantenerse una separación mínima de 20 cm entre la antena de este dispositivo y las personas durante su
utilización. Para garantizar dicho cumplimiento, no se recomienda su utilización a una distancia inferior a esta. La(s) antena(s)utilizada(s) para este transmisor no debe(n) situarse ni utilizarse junto a otra antena o transmisor.
Las modificaciones no aprobadas expresamente por Elcometer Limited podrían anular la autorización del usuario a usar elequipo según las reglas de la FCC.
Este dispositivo cumple la(s) norma(s) RSS de exención de licencia de Industry Canada. Su utilización está sujeta a lassiguientes dos condiciones: (1) Este equipo no puede provocar interferencias, y (2) este equipo debe aceptar cualquierinterferencia recibida, incluidas las interferencias que puedan provocar un funcionamiento no deseado del dispositivo.
Este aparato digital de Clase B cumple la norma CAN ICES-3 (B)/NMB-3(B).
y ElcoMaster® son marcas comerciales registradas de Elcometer Limited, Edge Lane, Manchester,M43 6BU. Reino Unido.
Las marcas comerciales Bluetooth pertenecen a Bluetooth SIG Inc y han sido licenciadas para
Elcometer Limited.
Elcometer 456 Modelo S y T: Creado para iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPhone 4s, iPhone 4, iPad Air 2,
iPad mini 3, iPad Air, iPad mini 2, iPad (3ª y 4ª generaciones), iPad mini, iPad 2, y iPod touch (4ª y 5ª generaciones).
“Made for iPod”, “Made for iPhone” y “Made for iPad” indican que un accesorio electrónico ha sido diseñado para conectar
específicamente con iPod, iPhone o iPad, respectivamente, y ha obtenido del desarrollador el certificado de cumplimiento delas normas de funcionamiento de Apple. Apple no es responsable del funcionamiento de este dispositivo ni del cumplimiento
por parte del mismo de las normas de seguridad y de la normativa. Tenga en cuenta que el uso de este accesorio con iPod,
iPhone o iPad puede afectar al rendimiento inalámbrico.
iPad, iPhone y iPod touch son marcas comerciales de Apple Inc. registradas en EE.UU. y otros países.
App Store es una marca comercial de Apple Inc. registrada en EE.UU. y otros países.
Google Play es una marca comercial de Google Inc.
Todas las demás marcas comerciales se dan por reconocidas.
R

El Elcometer 456 se suministra en un paquete de cartón. Asegúrese de que este embalaje se desecha de forma respetuosacon el medio ambiente. Consulte a las autoridades locales en materia medioambiental para obtener información.
Sede central: Elcometer Limited, Edge Lane, Manchester, M43 6BU. Reino Unido

TMA-0488-04 emisión 17 - Texto con Cubierta 22373-04

R

Handleiding
Elcometer 456
laagdiktemeter

www.elcometer.com

R

INHOUD
Sectie1Overzicht
2
Doosinhoud3
Gebruik van de meter4
Aan de slag
5
Aansluiten van de sonde
6
Nemen van een meting
7
Meter kalibreren
8
De kalibratie vergrendelen en ontgrendelen
Kalibratie methodes
9
Meet methoden
1011Geheugen12
Grafieken weergeven
13
Menu structuur - Model T
14
Menu structuur - Model S
15
Menu structuur - Model B
16
Gegevens downloaden en de meter upgraden
17
Technische specificaties
18
Juridische kennisgevingen & wettelijke informatie

Pb

Pagina
nl-2
nl-2
nl-3
nl-3
nl-4
nl-4
nl-5
nl-5
nl-6
nl-7
nl-8
nl-8
nl-10
nl-11
nl-12
nl-13
nl-14
nl-15

Hg

Om fouten en twijfel te voorkomen, gebruik a.u.b. de originele Engelse handleiding.
De Elcometer 456 is beschikbaar in 3 modellen. Deze handleiding is geschreven voor model T. Waar van
toepassing worden Model B en Model S genoemd.
Meter afmetingen:141 x 73 x 37mm (5,55 x 2,87 x 1,46'').
Meter gewicht:Geïntegreerde sonde: 156g (5,5oz) inclusief batterijen;
Afzonderlijke sonde: 161g (5,68oz) inclusief batterijen.
Toepasselijke patenten: US6243 661; US5886522; US6762603; US7606671; GB2306009; Gb2367135;
GB2342450, DE10131827© Elcometer Limited 2012 - 2018. Alle rechten voorbehouden. Niets van dit document mag worden
gereproduceerd, overgedragen, getranscribeerd, opgeslagen (in een retrieval systeem of anderszins) ofvertaald in enige taal, in welke vorm of door enig middel (elektronisch, mechanisch, magnetisch, optisch,handmatig of anderszins) zonder de voorafgaande schriftelijke toestemming van Elcometer Limited.

nl-1

www.elcometer.com

R

1 OVERZICHT
6

1
2

7
3
8

4

5

1
2
3
4
5
6
7
8

LED indicatoren - Rood (links), Groen (rechts)
Kleurenscherm
Multifunctionele Softkeys
Aan / uit knop
Interne sonde / separate sonde connectie
USB data output stekker (onder klepje)
Batterij compartiment (¼ slag open / dicht)
Polsband2 DOOSINHOUD











Elcometer 456 laagdiktemeter
Kalibratie folies (geïntegreerde meter)
Polsband
Beschermtas (B, S & T modellen)
Transporttas (T model)
1 x LCD scherm beschermer (S & T modellen)
2 x AA Batterijen
USB kabel & ElcoMaster® software (S & T)
Test certificaat
Handleiding

www.elcometer.com

nl-2

R

3 GEBRUIK VAN DE METERab

Batterij Status Indicator
Bluetooth aan - Grijs: nietaModel o
BST
ST
n

b
c
d
e
f
g

gekoppeld; Oranje: gekoppeldcd

Substraat type - F, N, FNF
Kalibratie methode

BST
BST

e

Meting Buiten Kalibratie
Waarschuwing Aan
Hoge Limiete Aan
Meeteenheden -

T

f
g

ST
BST

m

h

l

i
j

k

µm, mils, mm, inchh

Groep Type -pSTnormaal, geteld gemiddelde, IMO

i
j
k
l
m
n

Menu Softkey
Display Softkey
Groep / Data Softkey
Kalibrate Softkey
Meting Waarde
Gebruiker Selecteerbare
Statistieken - 4 rijenoGroep naam (wanneer in groeperen) T
Datum & Tijd (wanneer niet in groeperen) ST
BST
Staafgrafiek - hoogste, laagste en

p

BST
BST
BST
BST
BST
BST

q

r

s

gemiddelde waarde van de metingqrST
Lage Limiet Aan
Meet methoden - standaard, auto Ttherhaal en scan modusstGrafiek - laatste 20 metingen
Hoge & lage Limieten Aan

STST

4 AAN DE SLAG
4.1 ZORG VOOR DE LAATSTE FIRMWARE OP UW METER &UPGRADE UW METER
Om te verzekeren dat uw meter de laatste versie van de meter
firmware heeft – zodat u gebruik kunt maken van de laatste functies
en functionaliteit – adviseren wij dat de meter regelmatig wordt
verbonden met ElcoMaster® en voordat de meter voor de eerste keer
gebruikt wordt.

nl-3

www.elcometer.com

R

4 AAN DE SLAG (vervolg)
De meter via USB verbinden met een PC met internet en via
ElcoMaster® de optie 'Meter Verbinden' selecteren. Als een latereversie van de meter firmware beschikbaar is, zal 'Meter Update'verschijnen aan de rechterkant van meterdetails. Klik op 'Meter
Update' om de laatste firmware te installeren.
4.2 EEN TAAL SELECTEREN
1Druk en houd de aan / uit knop ingedrukt totdat het Elcometerlogo verschijnt.
2
Selecteer uw taal door middel van  softkeys.
3
Volg de menus op het LCD scherm.
Om het taal menu te bereiken als een vreemde taal is ingeschakeld:
1
Zet de meter uit.
2
Druk en houd ingedrukt de linker softkey en zet de meter aan.
3
Selecteer uw taal door middel van  softkeys.
5 AANSLUITEN VAN UW SONDEALLEEN VOOR ELCOMETER 456 MET SEPARATE SONDE
1
Draai de sonde om de pinnen op lijn te zetten.
2
Draai de sonde vast - met de klok mee.

2

1

2

1

6 NEMEN VAN EEN METING
1
2
3

Houd de sonde vast bij zijn omhulsel.
Meten, zet de sonde verticaal op het oppervlak.
Vervolg metingen, neem de sonde van het oppervlak en zet dezedaarna weer op het oppervlak.

www.elcometer.com

nl-4

R

6 NEMEN VAN EEN METING (vervolg) DOEN•
Houd de sonde vast bij het uiteinde.
•
Plaats de sonde voorzichting op het oppervlak.
•
Laat het uiteinde van de sonde contact maken met het
oppervlak - om de accuratesse van de meting te verbeteren.
 NIET DOEN•
Schuif de sonde over het oppervlak met coating.
•
Hard neerzetten van de sonde op het oppervlak.
•
De sonde boven het oppervlak laten ''zweven'', dit kan
resulteren in een foutieve meting.
LCD scherm verlichting wordt verminderd als er 15 seconden geen gebruik
wordt gemaakt van de meter en gaat helemaal uit als er geen gebruik wordt
gemaakt voor een periode zoals gedefinieerd in Menu/Setup/Scherminstellingen/Scherm uit - tijd. Druk op een knop of raak de meter aan om deze te activeren.
De meter sluit automatisch af als deze 5 minuten niet gebruikt is.
- - - geeft aan dat de meting buiten bereik van de sonde is.

7 KALIBREREN VAN DE METER
1
Druk de Kal softkey.
2
Voor alternatieve kalibratie methoden,
selecteer Kal/Kal Methode.
3
Selecteer Kalibreren en volg de
instructies op het LCD scherm.
4
Wanneer gevraagd plaats de sondein het midden van de folie(s).



90°

Folies kunnen op elkaar worden geplaatst.
Niet alle kalibratie methoden zijn beschikbaar voor all meter typen zie 9
Kalibratie Methoden

8 DE KALIBRATIE VERGRENDELEN EN ONTGRENDELEN
Een pincode voor kalibratievergrendeling instellen:
1
Druk op de softkey Menu en kies
Instellingen/Kalibratie vergrendelen.
2Stel een pincode van vier karakters in
met de softkeys  om 0 tot 9
te kiezen en met de softkey  om
van het eerste naar het vierde
karakter† te bewegen.
3
Druk op OK, Escape om de pincode teannuleren of op Aanpassen om depincode te wijzigen.
†

De softkey  verschijnt als de "X" verandert in een getal.

nl-5

www.elcometer.com

R

8 DE KALIBRATIE VERGRENDELEN EN ONTGRENDELEN (vervolg)
De kalibratie ontgrendelen:
1
Druk op de softkey Cal en kies Kalibratie vergrendelen.
2
Voer de pincode van vier karakters in, indien ingesteld, met desoftkeys  om 0 tot 9 te kiezen en met de softkey  om van
het eerste naar het vierde karakter† te bewegen.
3
Druk op OK of Escape om te annuleren.
De pincode voor kalibratievergrendeling deactiveren:
1
Druk op de softkey Menu en kies Instellingen/Kalibratie vergrendelen.
2
Voer de pincode van vier karakters in.


Als u de pincode bent vergeten of kwijtgeraakt, kunt u de pincode deactiveren
via ElcoMaster®. Sluit de meter met de USB-kabel aan op een pc met daarop
ElcoMaster® versie 2.0.33 of nieuwer geïnstalleerd en kies Bewerken/Wis
kalibratie pincode.

9 KALIBRATIE METHODES†KALIBRATIEMETHODEMODELOMSCHRIJVING

Nul

B, S, T

Een punt kalibratie - ideaal voor gladde
oppervlakken, plaats de sonde op het oppervlak(zonder coating) en de meter kalibreert
automatisch.

Glad

B, S, T

Twee punts kalibratie - de gebruiker selecteert
een bruikbare folie en een glad oppervlak(zonder coating = nul).

Ruw / 2 punts

B, S, T

De ideale kalibratie methode voor ruwe
oppervlakken, gebruik 2 folies met bekende
diktes - één boven en één onder dewaarschijnlijke laagdikte.

Nul offset

S, T

Kalibratie methode voor metingen waarbij het
oppervlakte profiel / ruwheid niet bekend is of niet
bereikbaar. Een door de gebruiker bepaalde offset
waarde wordt toegepast op de gemeten waarde.

De softkey  verschijnt als de "X" verandert in een getal.

www.elcometer.com

nl-6

R

9 KALIBRATIE METHODES (vervolg)
KALIBRATIEMETHODEMODELOMSCHRIJVING

Auto

S, T

Ideaal voor terugkerende inspecties. De gebruiker
slaat de bekende kalibratie folie dikte op in demeter. Wanneer de meter wordt gekalibreerd zal
de gebruiker worden gewaarschuwd om de in het
geheugen opgeslagen waarden te gebruiken - het
kalibratie proces wordt hierdoor vereenvoudigd enversneld.

ISO

S, T

Selecteert de kalibratie methode naar Nul offset
en zet het geteld gemiddelde op 5 inovereenstemming met ISO19840.

SSPC PA2

S, T

Selecteert de kalibratie methode naar Ruw / 2
punts en zet het geteld gemiddelde op 3 inovereenstemming met SSPC Pa2.

ZweedsS, T

Selecteert de kalibratie methode op Ruw / 2
punts en zet het geteld gemiddelde op 5 inovereenstemming met SS normen.

AustralischS, T

Selecteert de kalibratiemethode naar Nul
offset en zet het geteld gemiddelde op 5 inovereenstemming met AS normen.

10 MEET METHODEN (MODEL T)
10.1 MEET METHODEN
Drie meet methoden zijn beschikbaar: Standaard modus, Auto Herhaal
Modus en Scan Modus. Om de meet methode te selecteren, druk op
Menu/Setup/Meet methode.
‘Auto herhaal modus’ en ‘Scan modus’ zijn alleen beschikbaar met een Elcometer
456 Ultra / Scan sonde. Voor meer informatie contacteer Elcometer of bezoek
www.elcometer.com.

10.2 NAVSEA MODUS
Toont de meting bij de dichtstbijzijnde 1 mil (1 µm) boven 10 mils
(254 µm) en 0.1 mil (0.1 µm) onder 10 mils (254 µm).
Om NAVSEA modus te selecteren, druk Menu/Setup/NAVSEA of
Groep/Nieuwe Groep/NAVSEA.
nl-7

www.elcometer.com

R

11 GEHEUGEN (S & T MODELLEN)
1
Om het geheugen te gebruiken, druk de Groep softkey.
2
Selecteer ‘Nieuwe Groep’ of ‘Open Bestaande Groep’.
3
Kopieer en bekijk groep data.
4
Selecteer ‘Aanpassen Groep’ om ‘Hernoem Groep’, ‘Verwijder
Metingen’ of ‘Verwijder Groep’.
5
Vaste Groep Grootte laat gebruikers een van te voren bepaald
vast aantal metingen opslaan in een groep. Als alle metingen zijn
genomen zal de meter automatisch een nieuwe groep openen
met een link naar de originele groep naam. Bijvoorbeeld Nieuwe
Groep_1 wordt Nieuwe Groep_2, Nieuwe Groep_3, etc.
Sla elke meting op in het geheugen of sla het gemiddelde van een vooraf bepaalde
hoeveelheid metingen door middel van de geteld gemiddelde functie op.

12 GRAFIEKEN WEERGEVEN
12.1 STAAFGRAFIEKDe staafgrafiek toont een analoge
weergave van de dikte samen met dehoogste, laagste en gemiddelde waarde
wanneer metingen worden genomen. Om
de staafgrafiek te tonen:
1

Druk op de softkey Display en kies
‘Metingen en staafgrafiek’.



Als een meting buiten de ingestelde limieten valt, worden de witte grafiek en dewaarde van de meting rood.
Wanneer in ''Scan'' modus (Model T), wordt de staafgrafiek automatisch getoond
tijdens scannnen.

12.2 GRAFIEK (S & T MODELLEN)
Een grafiek van de laatste 20 metingen
tonen:
1
Druk de Groep softkey.
2
Selecteer ‘Nieuwe Groep’ of ‘Open
Bestaande Groep’.
3
Druk op de softkey Display en kies
‘Metingen en grafiek’.
Rode kolommen geven metingen aan die buiten de grenzen van de groep vallen(als deze zijn ingesteld).

www.elcometer.com

nl-8

R

12 GRAFIEKEN WEERGEVEN
12.3 GROEP GRAFIEK (MODEL T)
Grafiek met groepsmetingen tonen:
1
2



nl-9

Kies de gewenste groepsnaam vanuit
Groep/Groep Inzien.
Selecteer ‘Groep grafiek’.
Rode punten geven metingen aan die buiten
de grenzen van de groep vallen (als deze zijn
ingesteld).
Druk op de softkey Zoom+ gevolgd door  of  om individuele metingen in tezien.

www.elcometer.com

www.elcometer.com

Selecteer Statistieken
Alle PSPC Waarden○
Aantal Metingen○
Gemiddelde○Laagste Meting○
Hoogste Meting○Standard Afwijking○
Variatie Coeffient○Laagste Limiet Waarde○
Aantal Onder Limiet○
Hoogste Limiet Waarde○
Aantal Boven Limiet○
EIV○
NDFT Waarde○%>=NDFT○% 90-100% NDFT○IMO 90:10 Pass/Fail
○

○
○
○
○
○

waar n = 1, 2, of 3

Kal Geheugen n
Leeg hh:mm dd/mm/yy
SelectKalibreren
Kal Methode
Test Kalibratie
Hernoem Kal Geheugen n
Inzien Kalibratie Data
Verwijder Kal Geheugen n

Kal MethodeNul
Glad
Ruw / 2 Punts
Nul Offset
Auto

Statistieken
Stats Verwijderen
Inzien Geselecteerde
Alles Inzien
Selecteer Statistieken
Geteld Gemiddelde

Kalibratie
Kalibreren
Kal MethodeKal Geheugen 1
Kal Geheugen 2
Kal Geheugen 3
Fabrieks Kalibratie
Kalibratie Slot
Test Kalibratie
Kalibratie Check

Groep
Vrij Geheugen
Nieuwe Groep
Open Bestaande Groep
Verlaat Groeperen?
Aanpassen groep
Kopieer Groep
Groep Inzien
Meting Verwijderd

Display
Stats Verwijderen
Metingen
Metingen en statistieken
Metingen en grafiek
Metingen en staafgrafiek
Groep grafiek
Statistieken

Menu
Verwijder Laatste Meting
Setup
Limieten Instellen
BluetoothInfoOpnieuw Instellen
Noodverlichting

○

○

○
○
○
○

○

○

○

Meting Verwijderd
Verwijder Met Teken
Verwijder Zonder Teken

Groep Inzien
Groep 1
Statistieken
Groep Informatie
Kalibratie Informatie
Metingen
Groep grafiek

Aanpassen groep
Hernoem Groep
Verwijder Metingen
Verwijder Groep

Nieuwe Groep
Open Groep 1
Hernoem Groep 1
Groep Kalibratie
Groep Type
Groep Limieten
Vaste Groep Grootte
NAVSEA

Info
Meter Informatie
Sonde Informatie
Contact
Juridisch○○○Bluetooth
Aanzetten Bluetooth○Bluetooth ID
xxxxxxBluetooth CodexxxxxxAI

Limieten Instellen
Limieten Aan
Instellen Hoog
Instellen Laag
Instellen NDFT

Setup
Maateenheden
Taal
Tijd en Datum
Beep Volume
Scherminstellingen
Sonde
Meet modus
Meter Automatisch Uit
Kalibratie SlotNAVSEA

○
○
○
○

Sonde
Auto FNF
Ferro F
Non-Ferro N
Aanzetten sonde aan

Groep Limieten
Limieten Aan
Instellen Hoog
Instellen Laag
Instellen NDFT

Groep Type
NormaalGeteld Gemiddelde
IMO PSPC

Groep Kalibratie
Kalibreren
Kal MethodeGebruik Kal Geheugen 1
Gebruik Kal Geheugen 2
Gebruik Kal Geheugen 3

Patenten
Juridische aantekeningen
Regelgeving

JuridischKalibratie Slot
Pin code aanzetten

Meet modus
Standaard modus
Auto herhaal modus
Scan Modus
Scan modus met stop

○

○
○
○

○
○
○

○
○
○
○

○
○

○
○

Scherminstellingen
Helderheid Van Scherm
Scherm uit - tijd
Auto Rotatie Tonen
Openingsscherm

Sonde

○

Tijd en Datum
Tijd Instellen
Datum Instellen
Formaat Instellen
Toon Tijd en Datum

Ferro F1
Ferro F2○○○○

Maateenheden
µm
mm
mils
inch

Sonde

Regelgeving

Kal MethodeNul
Glad
Ruw / 2 Punts
Nul Offset
Auto
ISO
SSPC PA2
ZweedsAustralisch

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Menu is afhankelijk van
welke sonde wordt gebruiktR

13 MENU STRUCTUUR - ELCOMETER 456 MODEL T

nl-10

nl-11

Selecteer Statistieken
Alle PSPC Waarden○
Aantal Metingen○
Gemiddelde○Laagste Meting○
Hoogste Meting○Standard Afwijking○
Variatie Coeffient○Laagste Limiet Waarde○
Aantal Onder Limiet○
Hoogste Limiet Waarde○
Aantal Boven Limiet○
EIV○
NDFT Waarde○%>=NDFT○% 90-100% NDFT○IMO 90:10 Pass/Fail○Kal MethodeNul
Glad
Ruw / 2 Punts
Nul Offset
Auto

Statistieken
Stats Verwijderen
Inzien Geselecteerde
Alles Inzien
Selecteer Statistieken
Geteld Gemiddelde

○
○
○
○
○
Kalibreren
Kal MethodeFabrieks Kalibratie
Kalibratie Slot
Test Kalibratie

Kalibratie

Groep
Vrij Geheugen
Nieuwe Groep
Open Bestaande Groep
Verlaat Groeperen?
Aanpassen groep
Groep Inzien
Meting Verwijderd

Display
Stats Verwijderen
Metingen
Metingen en statistieken
Metingen en grafiek
Metingen en staafgrafiek
Statistieken

Menu
Verwijder Laatste Meting
Setup
Limieten Instellen
BluetoothInfoOpnieuw Instellen
Noodverlichting

○

○
○
○
○

○

○

Meting Verwijderd
Verwijder Met Teken
Verwijder Zonder Teken

Groep Inzien
Groep 1
Statistieken
Groep Informatie
Kalibratie Informatie
MetingenAanpassen groep
Verwijder Metingen
Verwijder Groep

Nieuwe Groep
Open Groep
Groep Kalibratie
Groep Type

Info
Meter Informatie
Sonde Informatie
Contact
Juridisch○○Bluetooth
Aanzetten Bluetooth○Bluetooth ID
xxxxxxBluetooth CodexxxxxxAI

Limieten Instellen
Limieten Aan
Instellen Hoog
Instellen Laag
Instellen NDFT

Setup
Maateenheden
Taal
Tijd en Datum
Beep Volume
Scherminstellingen
Sonde
Meter Automatisch Uit
Kalibratie Slot

○
○
○
○

Sonde
Auto FNF
Ferro F
No-Ferro N
Aanzetten sonde aan

Groep Type
NormaalGeteld Gemiddelde
IMO PSPC

Groep Kalibratie
Kalibreren
Kal MethodeJuridisch
Patenten
Juridische aantekeningen
Regelgeving

Kalibratie Slot
Pin code aanzetten

○
○
○

○
○

○
○

Scherminstellingen
Helderheid Van Scherm
Scherm uit - tijd
Auto Rotatie Tonen
Openingsscherm

Sonde

○

Tijd en Datum
Tijd Instellen
Datum Instellen
Formaat Instellen
Toon Tijd en Datum

Ferro F1
Ferro F2○○○○

Maateenheden
µm
mm
mils
inch

Sonde

Kal MethodeNul
Glad
Ruw / 2 Punts
Nul Offset
Auto
ISO
SSPC PA2
ZweedsAustralisch

QDID

Regelgeving

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

Menu is afhankelijk van
welke sonde wordt gebruiktR

14 MENU STRUCTUUR - ELCOMETER 456 MODEL S

www.elcometer.com

www.elcometer.com

Selecteer Statistieken
Aantal Metingen○
Gemiddelde○Laagste Meting○
Hoogste Meting○Standard Afwijking○
Variatie Coeffient○
EIV○

Kal MethodeNul
Glad
Ruw / 2 Punts

Statistieken
Stats Verwijderen
Inzien Geselecteerde
Alles Inzien
Selecteer Statistieken
Geteld Gemiddelde○○○

Kalibratie

Verwijder

Kalibreren
Kal MethodeFabrieks Kalibratie
Kalibratie Slot
Test Kalibratie

Terug

51.5µmGegevsLaatste Vijf Metingen
50.5 µm52.3 µm51.0 µm51.8 µm

Display
Stats Verwijderen
Metingen
Metingen en statistieken
Metingen en staafgrafiek
Statistieken

Menu
Verwijder Laatste Meting
Setup
Info
Opnieuw Instellen
Noodverlichting

○

Ok

○
○
○

Info
Meter Informatie
Sonde Informatie
Contact
Juridisch

Setup
Maateenheden
Taal
Beep Volume
Scherminstellingen
Sonde
Meter Automatisch Uit
Kalibratie Slot

○

Sonde

Patenten
Juridische aantekeningen
Regelgeving

JuridischKalibratie Slot
Pin code aanzetten

Ferro F1
Ferro F2

○
○

○
○
○
○

○
○

Scherminstellingen
Helderheid Van Scherm
Scherm uit - tijd
Auto Rotatie Tonen
Openingsscherm
Sonde
Auto FNF
Ferro F
No-Ferro N
Aanzetten sonde aan

○
○
○
○

Maateenheden
µm
mm
mils
inch

Sonde

Regelgeving
xxxxx

Menu is afhankelijk van
welke sonde wordt gebruiktR

15 MENU STRUCTUUR - ELCOMETER 456 MODEL B

nl-12

R

16 GEGEVENS DOWNLOADEN EN DE METER UPGRADEN
16.1 MET ELCOMASTER®
Bij gebruik ElcoMaster® - geleverd met de Elcometer 456
Model S & T, en beschikbaar als gratis download op elcometer.com alle meters kunnen meetgegevens overzetten voor opslag en het
maken van rapporten. Meetgegevens kunnen worden overgezet via
USB of Bluetooth® (S & T modellen). Ga naar www.elcometer.com voor
meer informatie over ElcoMaster®.
16.2 MET DE ELCOMASTER® APP (S & T MODELLEN)
Ideaal voor gebruik in het veld of op locatie. Met de ElcoMaster® App
voor Android™ of iOS kunt u:






Realtime metingen direct opslaan opmobiele apparatuur en in groepen
opslaan in combinatie met GPScoördinaten.
Foto's toevoegen van het
testoppervlak.
Metingen toewijzen aan een kaart, fotoof diagram.
De inspectiegegevens van mobiele
apparaten overzetten naar een pc omdeze verder te analyseren en er
rapporten van te maken.

Ga naar www.elcometer.com voor meer
informatie over de ElcoMaster® Apps.

Compatibel met smartphones entablets die draaien onder Android
2.1 of nieuwer. Installeer de app
door deze te downloaden in de
Google Play™ Store of via
www.elcometer.com en volg de
instructies op het scherm.

nl-13

www.elcometer.com

R

16 GEGEVENS DOWNLOADEN EN DE METER UPGRADEN (vervolg)
Geschikt voor iPhone 6 Plus,
iPhone 6, iPhone 5s, iPhone 5c,
iPhone 5, iPhone 4s, iPhone 4,
iPad Air 2, iPad mini 3, iPad Air,
iPad mini 2, iPad (3e en 4e
generatie), iPad mini, iPad 2, en
iPod touch (4e en 5e generatie).
Installeer de app door deze tedownloaden via de App Store of
via www.elcometer.com en volgde instructies op het scherm.
16.3 DE METERSOFTWARE UPGRADEN
Via ElcoMaster® kunt u de firmware van de meter bijwerken naar demeest recente versie zodra deze beschikbaar is. ElcoMaster®
informeert u over updates als de meter is aangesloten op een pc met
internettoegang.
17 TECHNISCHE SPECIFICATIES
Batterij type2 x AA batterijen, oplaadbare batterijen
kunnen ook gebruikt worden

Gebruiks-temperatuur

van -10 tot 50°C (14 tot 122°F)

Relatieve vochtigheid

0 tot 95%

Meter afmetingen
Meter gewicht(met geleverde batterijen)

14,1 x 7,30 x 3,70cm(5,55 x 2,87 x 1,46")Geïntegreerde sonde: 156g (5,5oz)
Afzonderlijke sonde: 161g (5,68oz)

Kan gerbuikt worden in overeenstemming met:
AS 2331.1.4, AS 3894.3-B, AS/NZS 1580.108.1, ASTM B 499, ASTM D 1186-B,
ASTM D 1400, ASTM D 7091, ASTM E 376, ASTM G 12, BS 3900-C5-6B,
BS 3900-C5-6A, BS 5411-11, BS 5411-3, BS 5599, DIN 50981, DIN 50984,
ECCA T1, EN 13523-1, IMO MSC.215(82), IMO MSC.244 (83), ISO 1461,
ISO 19840, ISO 2178, ISO 2063, ISO 2360, ISO 2808-6A, ISO 2808-6B,
ISO 2808-7C, ISO 2808-7D, ISO 2808-12, NF T30-124, SS 184159, SSPC PA 2,
US Navy PPI 63101-000, US Navy NSI 009-32

www.elcometer.com

nl-14

R

18 JURIDISCHE KENNISGEVINGEN & WETTELIJKE INFORMATIE
Conformiteitsverklaring:
Elcometer 456 Model B voldoet aan de vereisten van de volgende EU Richtlijnen:
2014/30/EU
Elektromagnetische Verenigbaarheid
2011/65/EU
Beperking voor het gebruik van bepaalde gevaarlijke stoffen
Elcometer 456 Model S & T voldoen aan de vereisten van de volgende EU Richtlijn
2014/53/EU
Radioapparatuur
2011/65/EU
Beperking voor het gebruik van bepaalde gevaarlijke stoffen
Verklaringen van Conformiteit zijn beschikbaar om te downloaden via:
Model B: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration of Conformity/Dutch/DoC_456C_B.pdf
Modellen S & T: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration of Conformity/Dutch/DoC_456C_S&T.pdf
Operationele frequentieband: 2,402 - 2,480 MHz
Maximale zendvermogen: <4 dBm
CISPR 11 geclassificeerd als Klasse B, Groep 1 ISM apparaat.
Klasse B producten: zijn geschikt voor gebruik in huishoudens en ruimtes die aangesloten zijn op het openbare
laagspanningsnetwerk.
Groep 1 ISM producten: producten waarin opzettelijk geleidend gekoppelde radiofrequente energie wordt opgewekt ofgebruikt voor de interne werking van het apparaat zelf.
De USB-poort is alleen geschikt voor het overdragen van informatie en mag niet met een adapter op de netvoeding worden
aangesloten.
Dit apparaat voldoet aan deel 15 van de FCC regels. Werking is onderheving aan de volgende 2 voorwaarden. (1) Dit apparaat
mag geen kwalijke storingen veroorzaken, en (2) dit apparaat moet storingen qua ontvangst kunnen verwerken, inclusief storingen
die zouden kunnen resulteren in het niet behoorlijk functioneren van het apparaat.
Het ACMA-keurmerk kunt u vinden in: Menu/Info/Juridisch/Regelgeving
Elcometer 456 modellen S & T: Het Giteki-symbool, ordinantienummer, FCC ID en Bluetooth SIG QDID kunt u benaderen via:
Menu/Info/Juridisch/Regelgeving
OPMERKING: Dit apparaat is getest en voldoet aan de limieten voor een Klasse B digitaal apparaat, conform Deel 15 van de FCC
Regels. Deze limieten zijn ontworpen om een redelijke bescherming te bieden tegen kwalijke storing in een huisinstallatie. Dit
apparaat genereert en gebruikt radiofrequente energie en kan die uitstralen. En als het apparaat niet wordt geïnstalleerd en gebruikt
volgens de gebruiksaanwijzing kan het kwalijke storing aan radiocommunicatie veroorzaken. Het is echter geen garantie dat er in
bepaalde installaties geen storing kan voorkomen. Als dit apparaat kwalijke storing veroorzaakt aan radio- of televisieontvangst, wat u
kunt vaststellen door het apparaat in- en uit te schakelen, wordt u aangeraden om te proberen om de storing te verhelpen d.m.v. eenof meerdere van de volgende maatregelen:-- Herschikken of verplaatsen van de ontvangstantenne.
-- De afstand tussen het apparaat en de ontvanger vergroten.
-- Het apparaat aansluiten op een andere groep dan die waarop de ontvanger is aangesloten.
-- De verkoper of een ervaren radio/tv-monteur raadplegen.
Om te voldoen aan de FCC-eisen voor RF-blootstelling bij mobiele en vaste zendapparatuur dient men tijdens bedrijf minimaal
20 cm afstand te houden tot de antenne van dit apparaat. Om naleving te garanderen, raden we u aan deze afstand te
respecteren. De antenne(s) die wordt/worden gebruikt voor deze zender niet samenvoegen of gebruiken in combinatie met
andere antennes of zenders.
Door modificaties uit te voeren die niet uitdrukkelijk zijn goedgekeurd door Elcometer Limited kan gebruik van het apparaat
buiten de FCC-reglementen vallen.
Dit apparaat voldoet aan de eisen van Industry Canada voor licentievrije RSS-standaard(en). De werking is onderhevig aan devolgende twee voorwaarden: (1) dit apparaat mag geen storingen veroorzaken, en (2) dit apparaat moet storingen kunnen
verwerken, inclusief storingen die zouden kunnen resulteren in het niet behoorlijk functioneren van het apparaat.
Dit Klasse B geclassificeerde digitale apparaat voldoet aan de CAN ICES-3 (B)/NMB-3(B) normen.
en ElcoMaster® zijn gedeponeerde handelsmerken van Elcometer Limited, Edge Lane, Manchester,M43 6BU. Verenigd Koninkrijk.
zijn handelsmerken van Bluetooth SIG Inc waarvoor een licentie is verleend aan Elcometer Limited.
Elcometer 456 Modellen S & T: Geschikt voor iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPhone 4s, iPhone 4, iPad Air
2, iPad mini 3, iPad Air, iPad mini 2, iPad (3e en 4e generatie), iPad mini, iPad 2, en iPod touch (4e en 5e generatie).
Met de uitdrukkingen 'Made for iPod', 'Made for iPhone' en 'Made for iPad' wordt bedoeld dat deze elektronische accessoires
speciaal zijn ontworpen voor respectievelijk de iPod, iPhone of iPad en dat de ontwerper verklaart de prestatiestandaarden
van Apple te hebben nageleefd. Apple is niet verantwoordelijk voor de werking van dit apparaat of dat het voldoet aanveiligheidsstandaarden en wettelijke standaarden. Let op: als u deze accessoire gebruikt in combinatie met een iPod, iPhoneof iPad kan dit de prestaties van de draadloze verbinding beïnvloeden.
iPad, iPhone en iPod touch zijn handelsmerken van Apple Inc., gedeponeerd in de VS en andere landen.
App Store is een handelsmerk van Apple Inc., gedeponeerd in de VS en andere landen.
Google Play is een handelsmerk van Google Inc.
Alle andere handelsmerken zijn het eigendom van hun respectievelijke eigenaars.
R

De Elcometer 456 is verpakt in karton. Zorg ervoor dat alle verpakking milieuvriendelijk wordt afgevoerd. Neem contact op
met de milieuafdeling van uw gemeente voor advies.
Hoofdkantoor: Elcometer Limited, Edge Lane, Manchester, M43 6BU. Verenigd Koninkrijk

TMA-0488-11 Uitgave 17 - Text met voorblad No. 22373-11

R

用户使用手册
易 456涂层测厚仪

www.elcometer.com

R

目录
部分

页

1

仪

2

述

zh-2

清单

zh-2

使用

zh-3

3

仪

4

启动

5

zh-3

头连
记录

zh-4

6
7

据

zh-4

仪

8

锁

zh-5

和

锁

zh-5
zh-6

9
10

测量模式

zh-7

11

据分

zh-8

12

显示图

13

菜单结构—

14

菜单结构—标

型（S）

zh-11

15

菜单结构—基本型（B）

zh-12

记录

zh-8

16

载

据

17

技术规格

级型（T）

仪

律提示 和

Pb

Hg

易

英

456涂层测厚仪有3

升级

zh-13
zh-14

18

疑议，请

zh-10

规

息

zh-15

本。
型

可选，本手册

据

级型（T）编

，同样

用于、基本型（B型）和标仪

尺寸：

141x73x37mm (5.55x2.87x1.46“)。

仪

重量：

整体式： 156g (5.5oz) 包括电池；分体式： 161g (5.68oz) 包括电池。

应用专利

:

US6243 661; US5886522; US6762603; US7606671; GB2306009; GB2367135; GB2342450, DE10131827© Elcometer Limited 2012 - 2018. 司保有权利。本中），或者在 有Elcometer Limited事先书面许可的情况
任何语 。

zh-1

型（S型）。

任何 分都不得复制、传输、存储（在检 或 检
统
任何 式（电子、机械、 性、光学、手动或其他）译成

www.elcometer.com

R

1仪

述
6

1
2

7
3
4

8

5

1

LED指示灯-红灯（左边），绿灯（右边）

2
3
4
5
6
7
8

彩屏显示
多功
键
开/关 键
头/分体 头连
USB 据输 插孔（在机盖电池盒（可打开/关闭）
带连

2

清单

）

易 456涂层测厚仪
膜片（整体式仪 ）带保护 （基本型、标 型
级型）
手提箱（ 级型）
1个屏幕保护 （标 型和 级型）

2节干电池

USB线 ElcoMaster® 软件（标 型和

检验证书

用户使用指
www.elcometer.com


级型）

zh-2

R

3仪

使用
型
BST

a

电池使用

b

d

蓝 开启功 - 色：不配对;橙 ST
色：配对基体类型（铁基、 铁基、两用型） BST
BST

e

读

c

命指示图标

T

报功

f

上限

g

j

测量单 — 米、 耳、 米、 BST
英尺ST
分 记录 类—正常、计算平
值、IMO
BST
菜单 键
BST
显示软键

k

分

h
i

ST

记录

l

据

BST

键

键

BST

m 读 值
n 用户可选统计—4

BST

o

据组名

（分

a
o
b
c
n

d
e
f
g

m
l

h

据组

显示

菜单

k

p

q

r

据组

显示

菜单

s

BST
记录情况

）

i
j

T

时间（不是分 记录情况 ） ST
BST
柱状图—
，
和平 读
日期

p
q

值设

r

测量模式—标

s

趋势图—

t

上

t

ST
,自动重复,扫描

后20个读

值开启功

T

据组

显示

菜单

ST
ST

4 启动
4.1
保您的仪
有
的固件和升级你的仪为 保您的仪 拥有
的固件，让您从
的 性和功
们 议仪
期连
ElcoMaster®和 一次使用前.

中受，我

只需使用“连 仪 ”功 ,通过USB连 仪
ElcoMaster® 联网连的电脑. 如果仪 固件的后续 本可 提供, “更 仪 ” 显示在仪 详
细资料右侧. 点击“更 仪 ”来
的固件.
zh-3

www.elcometer.com

R

4 启动（续前节）4.2 选择语
1开关键并保
2
用  键选择语3
据屏幕菜单操作仪屏幕显示“Elcometer“图标后，仪当选用 语时，进 语 菜单1
关闭仪2
左边的软 键并 续一
3
用  键选择语
5连
只
1
2

开启

时间，打开仪

头
用于分体式测厚仪转 头插头 仪顺时针 向拧紧 头

对齐

2

1

2

1

6读
1
2
3

住 头
获 读 时
头
在 测 面
用户需继续测量读 时，从 测基体带有涂层的基体 面 可



操作:
住 头保护
头轻轻 在 测 面
让 头保护
基体——达

•
•
•

www.elcometer.com

提

面提起

头，然后再

其

性的目的
zh-4

R

6读

（续前节）

 不要做:•••

在涂层 面拖拽 头用力头 在涂层 面头在涂层 面来回移动，这样测

如果不活动的时间超过15 ，显示屏 变 ，并 继续变'黑'，如果在不活动期间中 义的菜单/设 /屏幕设 /屏幕超时。请 任何键，或点击唤醒
5分钟 有任何操作，仪
自动关机- - - 示读 超过 头测量范围




7

仪

1
2
3
4

键
选择
，
键→
选择
键， 据屏幕提示操作
当屏幕 现提示，
头 在
片的中央

4


膜片可 叠加
不是 有

都可用在

有型

膜

据不

90°

的易 456涂层测厚仪，见 9页

8锁

和

设
1
2

锁 PIN码:菜单键，选择设 /锁设PIN码，使用  键选†
择0-9，  键移动一OK键，Escape键 消或Adjust键
PIN码

3

的

锁

锁
1
2
3
†

输
一
OK键

当x变成

zh-5

键，选择
锁
PIN码，如果设

时，，使用  键选择0-9，  键移动†

或Escape键消现键

www.elcometer.com

R

8锁

和

锁

（续前节）

禁用
锁 PIN码:1
菜单键，选择设 /
2
输
PIN码
4

锁

一旦用户 记或丢失PIN码，可通过ElcoMaster® 软件PIN码 禁用。使用USB
线，可轻 地 仪带ElcoMaster® 2.0.33 或更本的电脑连 ，选择编辑或
清除
PIN码

9

点

描述

B, S, T单点
头 在— 用于平 基体 面，
有涂层的基体上，仪自动—用户选择合涂层基体

平基体

B, S, T

两点
平 、

粗

基体/两点

B, S, T

用于粗 或 型基体, 用两个已知厚的膜片, 一个 于, 另一个 于 测涂
层的厚

S, T

用于重复测量，用户 已知
膜片厚 值存 仪 ，当开
时，用户
据屏幕提示
仪 ，仪
自动调
整存 的膜片厚 值——
过程简
单、快捷

自动

†

型

当x变成

时，

的

膜片和

现键

www.elcometer.com

zh-6

R

9

（续前节）
型

ISO

描述

S, T

设 为 点 移，设值为5— 据ISO19840标

平计算

SSPC PA2

S, T

设
算平

为粗 /两点，设
值为3— 据SSPCPA2标

计

瑞典标

S, T

设
算平

为粗 /两点值为5— 据瑞典标

计

S, T

设

大利

标

值为5—

，设

为 点 移，设
据 大利亚标

计算平

10 测量模式 ( 高级型 T )10.1 测量模式
三 测量模式可供选择:"标 模式","自动重复模式"和"扫描模式". 要选择测量模式， 菜单/设 /测量模式.

“自动重复模式”和“扫描模式”仅适用于Elcometer 456 Ultra/扫描探头。如需进一步信息，请联系Elcometer或访问
www.elcometer.com

10.2 NAVSEA 模式
显示读
近的1mil(1μm)10mils
10mils (254μm).

上 (254μm)和0.1mil (0.1μm)低于

选择NAVSEA模式, 菜单/设 /NAVSEA或 数据组/新建数据组/NAVSEA.

zh-7

www.elcometer.com

R

11 分
1
2
3
4
5



记录

据 (标

型S &

级型 T)

使用 据组记忆功 ，“分 记录”键选择创的 据组或打开已有 据组进行读复制和审组 据选择“编辑 组”进行重命名，删除读 或删除 组据组 据 量可由用户预先设 ，当 据组中的 量达 了预值后，仪 后自动一个 的 据组进行读 ，并和原有 据组
连 ，例如
据组 有
据组_1,
据组_2等.
一个 据存
据平 值

仪

的记忆库或通过计算平值功，存储预先设

据组中

12 显示图

12.1 柱状图
柱状图显示一个模拟的厚 值 测量，
和平 读示。要显示柱状图：1显示软键，选择“读如果读 超 设在“扫描”模式（型

和柱状图”
据组

显示

菜单

据组

显示

菜单

制，白色的柱状和读 值 变成红色T），柱状图自动显示在 个扫描

12.2 趋势图 ( 标 型 S & 级型 T )要显示趋势图 的 后20个读 ：
1
2
3

据组键
选择
据组或打开现有 据组按显示软键，选择“读 和趋势图 ”红点表示批次极限外的一个读（如果设置）

www.elcometer.com

zh-8

R

12 显示图

（续前节）

12.3 组图 (级型 T )若要显示 组图 :
1
2




zh-9

从 据组/审
名
选择" 组图"

据组选择

当的

据

红色柱状表示批次极限以外的一个读（如果设置）通过  变 键检阅 需的单个
读www.elcometer.com

www.elcometer.com

选择统计 据有PSPC值
读 个
平 值
值
值
标
变
极
值
于 值的读 个极
值于 值的读 个易 给 值NDFT值%大于或等于NDFT% 90-100% NDFTIMO 90:10通过/失败

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

/两点
移

检验
重命名
看
清除

选择

○
○
○
○
○

其中 n = 1,2 或 3

记忆 n
据
记忆 n

值

记忆 n
存储 hh:mm dd/mm/yy

自动

平
粗

统计
清除统计 据看选 的 据看 有 据选择统计 据
计算平 值

显示

记忆1
记忆2
记忆3
工厂
锁
检验
值
检

存
据组
打开现有 据组
退
据组？
编辑 组
复制 据组
回顾 据组
删除的读

可用

据组

○

○

○
○
○
○

息
息
式

关于

蓝

值

息

息

删除的读
删除带标签
删除不带标签

读
批组图

统计
据组

组

回顾 据组
据组 1

重 命名
清除 据组
删除 据组

编辑
据组

据组
打开 据组1
重 命名 据组 1
据组
据组类型
据组 值
固
据组 量
NAVSEA

律

联

头

仪

启动蓝
蓝
证
蓝
码

开启 值功
设
值
设
值
设 NDFT设

○○○xxxxxxxxxxxx
AI

○

○

○

○

单

开启 值功
设
值
设
值
设 NDFT

律

锁

据组

据组

值

据组类型

记忆 1
记忆 2
记忆 3
正常
计算平 值
IMO PSPC

使用
使用
使用

专利
法律提示
规 的

启用PIN码

○

○
○
○

○
○
○

○
○
○
○

测量模式
标 模式
自动重复模式
扫描模式
Hold 键可用的扫描模式

○
○
○
○
○
○

头

头

○
○

○

○
○
○
○

铁基F1
铁基F2

自动 FNF
铁基F
铁基 N
头启动

屏幕
屏幕超时
显示自动 转
打开显示屏

屏幕设

时间和日期
设 时间
设 日期
设 格式
显示时间和日期

µm
mm
mils
inch

/两点
移
自动
ISO
SSPC PA2
瑞典标
大利亚标

平
粗

QDID

规

菜单显示

头

的

由连

的

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

头决

13 菜单结构—易

清除统计 据只有读
读 和 选统计量
读 和趋势图
读 和柱状图
组图
统计

删除
设
设
值
蓝
关于
重 设
应急灯

菜单
后一个读

单
语
时间和日期
音量
屏幕设
头
测量模式
仪 自动关闭锁
NAVSEA设R

Elcometer456高级型仪

zh-10

zh-11选择统计 据有PSPC值
读 个
平 值
值
值
标
变
极
值
于 值的读 个极
值于 值的读 个易 给 值NDFT值%大于或等于NDFT% 90-100% NDFTIMO 90:10通过/失败

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

/两点
移

自动

平
粗

○
○
○
○
○

检验

锁

工厂
值

存
据组
打开现有 据组
退
据组？
编辑 组
回顾 据组
删除的读

可用

据组

○

息
息
式

组

息

删除的读
删除带标签
删除不带标签

读

编辑

据组

关于

回顾 据组
据组 1

据组
据组

息

统计
据组

清除
删除

打开 据组1
据组
据组类型

律

联

头

仪

蓝

值

○
○

xxxxxxxxxxxx
AI

○

○

○

单

正常
计算平 值
IMO PSPC

专利
法律提示
规 的

启用PIN码

铁基F1
铁基F2

自动 FNF
铁基F
铁基 N
头启动

据组类型

据组

律

锁

头

头

屏幕设
屏幕
屏幕超时
显示自动 转
打开显示屏

时间和日期
设 时间
设 日期
设 格式
显示时间和日期

µm
mm
mils
inch

○
○
○

○
○

○
○
○
○

○
○

○

○
○
○
○

/两点
移
自动
ISO
SSPC PA2
瑞典标
大利亚标

平
粗

QDID

菜单显示

头

规

的

由连

的

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

头决

Elcometer456标

清除统计 据看选 的 据看 有 据选择统计 据
计算平 值

统计

○
○
○
○

启动蓝
蓝
证
蓝
码

开启 值功
设
值
设
值
设 NDFT设设

14 菜单结构—易显示清除统计 据只有读
读 和 选统计量
读 和趋势图
读 和柱状图
统计菜单
删除 后一个读
设
设
值
蓝
关于
重 设
应急灯

单
语
时间和日期
音量
屏幕设
头
仪 自动关闭
锁R型仪

www.elcometer.com

www.elcometer.com

读 个
平 值
值
值
标
变
易 给

值

选择统计

据

○
○
○
○
○
○
○

平
粗

/两点

清除统计 据看选 的 据看 有 据选择统计 据
计算平 值

统计

○
○
○
检验

锁

工厂

退

值

删除

51.5µm据后5个读
50.5 µm52.3 µm51.0 µm51.8 µm

○

OK

○
○
○
律

联

头

仪

息
息
式

单
语
音量
屏幕设
头
仪 自动关闭
锁
关于

设

○

单

专利
法律提示
规 的

启用PIN码

铁基F1
铁基F2

自动 FNF
铁基F
铁基 N
头启动

律

锁

头

头

屏幕设
屏幕
屏幕超时
显示自动 转
打开显示屏

µm
mm
mils
inch

○
○

○
○
○
○

○
○

○
○
○
○

菜单显示

头

规

的

由连

的

xxxxx

头决

15 菜单结构—易显示清除统计 据只有读
读 和 选统计量
读 和柱状图
统计菜单
删除 后一个读
设
关于
重 设
应急灯

R

456基本型仪

zh-12

R

16

载

据

仪

升级

16.1 使用ELCOMASTER®
使用易 Elcometer456标 型和 级型提供的ElcoMaster® 软件或从易网 www.elcometer.com上免费 载该软件， 有易 456测厚仪能据传输 电脑上保存、生成报告， 据通过 USB或蓝
Bluetooth®传输 (标 型和 级型)。有关ElcoMaster®更多信息,访问
www.elcometer.com。
16.2 使用ELCOMASTER®手机应用程 （型 S和T）
在实地或在现场的理想测试, 使用ElcoMaster®Android™或iOS 移动应用
程序, 用户可以:






存储实时读
移动设备上, 并
们保存
次连同全球
统坐
标.
可 加 测试 面的照片.
地图上的读地图，照片或图 .
检测 据可从手机传送 电脑进行进一的分析和报告.

有关ElcoMaster®移动应用程
问www.elcometer.com

更多

息,访

兼容智能手机和运行Android2.1或以上的平板电脑。使用Google
Play™Store应用程序下载安装，并
按照屏幕上的说明。

zh-13

www.elcometer.com

R

16

载

据

仪

升级（续前节）这是为 iPhone 6 Plus, iPhone 6,
iPhone 5s, iPhone 5c, iPhone 5,
iPhone 4s, iPhone 4, iPad Air 2,
iPad mini 3, iPad Air, iPad mini 2,
iPad（ 3和 4代），iPad mini,
iPad 2, 和iPod touch（ 4和5代）制成。要安装，通过
www.elcometer.com下载或通过
App Store下载安装，然后按照屏
幕上的说明。

16.3 提升你的仪 （B,S,T型）
通过ElcoMaster®仪 的固件用户可 升级
本.
当仪 连拥有 联网的电脑，ElcoMaster® 通知您任何的更 .
17 技术规格
2节干电池，也可使用

电池类型

电电池

操作温

-10 - 50°C (14 - 122°F)

相对湿

0 - 95%

尺寸 x宽x长

14.1 x 7.30 x 3.70cm(5.55 x 2.87 x 1.46")

电池在

整体式: 156g (5.5oz)
分体式: 161g (5.68oz)

的重量从如 标 :
AS 2331.1.4, AS 3894.3-B, AS/NZS 1580.108.1, ASTM B 499,
ASTM D 1186-B, ASTM D 1400, ASTM D 7091, ASTM E 376,
ASTM G 12, BS 3900-C5-6B, BS 3900-C5-6A, BS 5411-11, BS 5411-3,
BS 5599, DIN 50981, DIN 50984, ECCA T1, EN 13523-1,
IMO MSC.215(82), IMO MSC.244 (83), ISO 1461, ISO 19840, ISO 2063,
ISO 2178, ISO 2360, ISO 2808-6A, ISO 2808-6B, ISO 2808-7C,
ISO 2808-7D, ISO 2808-12, NF T30-124, SS 184159, SSPC PA 2,
US Navy PPI 63101-000, US Navy NSI 009-32

www.elcometer.com

zh-14

R

18

律提示 &

规

息

符合性声明:
Elcometer 456 型 B符合以下 盟指 的要求
2014/30/EU
电 兼 性
2011/65/EU
制使用某些有 物质Elcometer 456 型 S 和 T符合以下 盟指 的要求
2014/53/EU
线电设备
2011/65/EU
制使用某些有 物质
符合性声明可通过网址 载：型 B: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/Chinese/DoC_456C_B.pdf
型 S & T: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/Chinese/DoC_456C_S&T.pdf
操作频 ：2,402 - 2,480 MHz
大传输功 ： <4 dBm
据CISPR 11, 该产品是B级, 1组ISM设备.
B级产品: 为国 机构 使用,
连为作住 用的 筑物提供的 压供电网络.
1组ISM产品：A类产品产生的/或使用的导电 合 频 量, 是设备
本 运作 必需的.
USB是用于 据传输而不可 通过USB电源 配 连电源.
该设备符合FCC法规 十五 规 ，操作时有 面两 情况：（1）本仪 不 产生有 干 ；（2）本仪 可
受
干 ， 响 仪 的使用。
ACMA遵守标志可以通过以下浏览：菜单/关于/ 律/规 的
Elcometer 456 型 S 和 T: Giteki标记， 例 码,FCC ID和Bluetooth蓝

SIG QDID 可以通过

/规 的.
备注：易

设备规

的

456涂层测厚仪已得

设备提供合理保护、使用该设备，可检测，符合FCC规
止有

对

干

的

中15章关于B类线电通讯造成有响。该设备产生，使用并发
干

。

此，易

司不

，这些规

线电频
保证在

浏览: 菜单/关于/的目的是为居住环

，如果用户
殊

要求

有
不

律

中

照操作说明
产生干

，如

果设备由于开关机对 音机或电视产生有 干 ，用户可尝试用
决问题：
-- 重
线
-- 大易 测厚仪设备 间的-- 不要 易 测厚仪设备连 在同一电路上-- 询易 销或经验 富的 线电专为了满足移动设备和基 发 设备的FCC RF规 要求，应保 该的 线和操作过程中人 人 间的20厘米 上的间.为 保合规性，不 议操作在 这个
更近. 线用于此发
不得在同一地点或 任何其他 线或发一起工作。
用户在 据FCC规
，使用易司 有在操作说明书提 的有关仪 调整事项，起操作失败。
此设备符合加拿大工业
照的RSS标 .操作应符合
两个 件：（1）本设备不 造成干 ，（2）本设备必须
受任何干 ， 括可 导
预期操作的干 。
B类设备符合CAN ICES-3 (B)/NMB-3(B)规 。
和 ElcoMaster®是Elcometer公司的注册商标, Edge Lane, 曼彻斯 , M43 6BU,英国。
蓝标 有权归蓝 SIG公司所有，易高公司得到蓝 SIG公司授权使用。
Elcometer 456 型 S & T: 这是为 iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPhone 4s, iPhone 4, iPadR

Air 2, iPad mini 3, iPad Air, iPad mini 2, iPad（ 3和 4代），iPad mini, iPad 2, 和iPod touch（ 4和 5代）制成。
“Made for iPod” ,“Made for iPhone” “Made for iPad”的意思是一个电子附件为专门连接到iPod，iPhone或iPad设计，分别和已经由开发者认证符合Apple性能标准.Apple不负责本装置或其符合安全和监管标准的操作. 请注, 在iPod,iPhone或iPad上使用此配件可能响 线性 .
iPad，iPhone和iPod touch是Apple Inc公司的注册商标，在美国和其他国家注册.
App Store是Apple Inc公司的注册商标，在美国和其他国家注册.
Google Play是 Google Inc 公司的商标.
有 标也都得 注册许可.
易高456是 在一个纸箱总.请

司： Elcometer公司的注册

保

有

环

感的

式处理. 请

询当地环

局为进一

指导.

标, Edge Lane, 曼彻斯 , M43 6BU,英国.

TMA-0488-10

本 17 -

封面

编22373-10

R

ユーザーガイド
Elcometer 456
膜厚計

www.elcometer.com

R

⽬次
セクション
1
本体外観
2
梱包内容
3
画⾯表⽰と機能
4
はじめに
5
プローブの接続
6
測定
7
膜厚計の調整8
調整値の固定と固定解除
9
調整⽅法
10
測定モード
11
読み取り値のバッチ保存
12グラフの表⽰13
メニュー構成-モデルT
14
メニュー構成-モデルS
15
メニュー構成-モデルB
16
ソフトウェアの使⽤とファームウェアのアップグレード
17
仕様
18
関連する法律と規制について

Pb

ページ
jp-2
jp-2
jp-3
jp-3
jp-4
jp-4
jp-5
jp-5
jp-6
jp-7
jp-8
jp-8
jp-10
jp-11
jp-12
jp-13
jp-14
jp-15

Hg

不明な点がある場合は、英語版の取扱説明書を確認してください。
Elcometer 456には3つのモデルがあります。このユーザーガイドは、モデルT⽤です。他の2つのモデル（B、S）に当てはまる部分には、その旨が記載されています。
本体⼨法：

141 x 73 x 37mm（5.55 x 2.87 x 1.46インチ） 。

本体重量（電池も含む）-⼀体型:156g(5.5オンス）、セパレート型：161g (5.68オンス) 。
該当する特許：US6243 661、US5886522、US6762603、US7606671、GB2306009、GB2367135、
GB2342450、DE10131827© Elcometer Limited 2012 - 2018. All rights reserved. この⽂書の⼀部または全部を、Elcometer Limitedの事前の書⾯による許可なく、いかなる形式や⽅法（電⼦的、機械的、磁気的、⼯学的、⼿動を問わず）によっても、複製、転送、保管（検索可能なシステムかどうかを問わず）、または他の⾔語に翻訳することを禁じます。

jp-1

www.elcometer.com

R

1 本体外観
6

1
2

7
3
8

4

5

1

LEDランプ-⾚（左）、緑（右）

2
3
4
5
6
7
8

カラー液晶画⾯
多機能ソフトキー
オン/オフボタン
内蔵プローブ/外付けプローブ接続部
データ出⼒⽤USB端⼦（カバーの下）
電池収納部（¼回転で開閉）
リストバンド取付部

2 梱包内容Elcometer 456膜厚計本体
校正⽤フォイル（⼀体型モデル）
リストバンド保護ケース（モデルB、S、およびT）
キャリーケース（モデルT）
スクリーンプロテクター1個（モデルSおよびT）
AA乾電池2本USBケーブルとElcoMaster®ソフトウェア（モデルSおよびT）
検査証明書



ユーザーガイド










www.elcometer.com

jp-2

R

3 画⾯表⽰と機能
a
b

モデル
BST o
電池残量インジケーター
Bluetooth - ペアリングしているとき ST
はオレンジ、していないときはグレー

c
d
e
f
g

BST
素地の種類 - F, N, FNF
BST
調整⽅法
調整範囲外の測定値の警告オン T
ST
上限値設定オン
測定単位- µm、mils、mm、インチ BST

h

バッチの種類 - 標準、読み取り値 ST

a
b
c

n

d
e
f
g

m
l

h

調整 バッチﾃﾞｨｽﾌﾟﾚｲ

メニュー

k

j

p

の平均、IMO

i
j
k
l
m
n
o

p
q
r

BST
メニューソフトキー
BST
ディスプレイソフトキー
BST
バッチ/データソフトキー
BST
調整ソフトキー
BST
読み取り値
ユーザーが選択可能な統計値 - 4個 BST
バッチ名（バッチ機能を使⽤しているとき） T
⽇付と時刻（バッチ機能を使⽤ ST
していないとき）
棒グラフ-最⼤、最⼩、平均膜厚値も表⽰ BST
ST
下限値設定オン測定モード - スタンダード、オー T

qr

調整 バッチﾃﾞｨｽﾌﾟﾚｲ

メニュー

s

t

トリピート、スキャンst

ランチャート - 最後の20個の読み取り値 ST
ST
上限値と下限値の設定オンi

調整 バッチﾃﾞｨｽﾌﾟﾚｲ

メニュー

4 はじめに
4.1 最新バージョンのファームウェアの確認とアップグレード
膜厚計の最新機能を利⽤できるように、最新版のファームウェアがあるかどうかをElcoMaster®で定期的に確認することをお勧めします。
このためには、インターネットに接続されているコンピュータに膜厚計をUSBで接続して、ElcoMaster®を実⾏します。ファームウェアの最新版がある場合は、膜厚計の詳細の右横に更新（Update
Gauge）ボタンが表⽰されます。このボタンをクリックして、最新のファームウェアをインストールしてください。
www.elcometer.com
jp-3

R

4 はじめに（続き）
4.2 ⾔語の選択
1
オン/オフボタンを押したまま、Elcometerのロゴが表⽰されるのを待ちます。
と ソフトキーを使って⾔語を選択します。
2
3
画⾯に表⽰される指⽰に従います。
日本語以外で表示されているときに、言語メニューにアクセスするには:
1
本体の電源を切ります。
2
左のソフトキーを押したまま、本体の電源を⼊れます。
と ソフトキーを使って⾔語を選択します。
3
5 プローブの接続
セパレート型のみ
1
プローブのプラグを、本体接続部のピンの並びに合わせます。
2
プローブを差し込んで右に回します。

2

1

2

1

6 測定
1
2
3

プローブのスリーブを持ちます。
塗装⾯と垂直になるようにプローブをゆっくり近づけて触れさせます。
続けて測定するには、⼀旦プローブを塗装⾯から離し、1回⽬と同じ要領で、測定箇所にプローブの先端を触れさせます。



正しい測定⽅法
プローブのスリーブを持って測定する。
塗装⾯にプローブをゆっくり近づける。
プローブの先端を塗装⾯に触れさせる（正確な測定値が得られ
ます）。

•
•
•

www.elcometer.com

jp-4

R

6 測定（続き）

 間違った測定⽅法•••

塗装⾯の上でプローブを引きずる。
プローブで塗装⾯をたたいたり強く押し付けたりする。
プローブをしばらく塗装⾯に近づけたままにする（間違って測定されます）。



何も操作せずに15秒経つと、本体の画⾯が暗くなります。メニュー→設定→画⾯の設定→スクリーンタイムアウトで指定した時間が経過すると、画⾯が真っ
暗になります。元の明るい画⾯に戻すには、キーをどれか押すか、画⾯を軽くタップしてください。
何も操作しないまま5分経つと、本体の電源が⾃動的に切れます。
画⾯に---と表⽰された場合は、プローブの測定範囲外です。



7 膜厚計の調整
1
2
3
4
4


調整ソフトキーを押します。
調整⽅法を変更するには、調整→調整⽅法を選択します。
［調整する］を選択し、画⾯に表⽰
される指⽰に従います。
画⾯に指⽰が表⽰されたら、フォイ
ルの中央にプローブを置きます。

90°

フォイルは積み重ねることができます。
モデルによって、使⽤できる調整⽅法が異なります。詳しくは、「9 調整⽅法」を参照してください。

8 調整値の固定と固定解除
調整値を固定するPIN（暗証番号）を設定
するには：
1
メニューソフトキーを押し、［設定］、［調整固定］の順に選択します。
と ソフトキーを使って0～9の番
2号を選択し、 ソフトキーを押して次の桁にカーソルを移動させ、4桁のPINを⼊⼒します。†
3
Okを押してPINを設定します。キャンセルする場合はEscape
を、PINを変更する場合は調節を押します。
†

画⾯の「X」の場所に番号を⼊⼒すると、 ソフトキーが表⽰されます。

jp-5

www.elcometer.com

R

8 調整値の固定と固定解除（続き）
調整値の固定を解除するには：
1
調整ソフトキーを押し、［調整固定］を選択します。
とソフトキーを使って0～9の番号を選択し、ソフトキーを2
押して次の桁にカーソルを移動させ、設定済みの4桁のPINを⼊⼒します。†
3
Okを押して固定を解除します。キャンセルする場合はEscapeを押します。
調整値を固定するPINを無効にするには：
1
メニューソフトキーを押し、［設定］、［調整固定］の順に選択します。
2
4桁のPINを⼊⼒します。


PINを忘れた場合は、ElcoMaster®を使ってPINを無効にすることもできます。このためには、ElcoMaster®バージョン2.0.33以上をインストールしているPCに、
USBケーブルで膜厚計を接続し、［Edit/ClearCalibrationPIN］を選択します。

9 調整⽅法
校正⽅法

モデル

説明

ゼロ

B, S, T

1点式調整-滑らかな⾯の測定時に最も適しています。塗装されていない素地の上にプローブを置くだけで、⾃動的に調整されます。

平滑⾯

B, S, T

2点式調整-適切な調整⽤フォイルと塗装されていない滑らかな素地（ゼロ値）を選択します。

粗⾯／２ポイント

B, S, T

粗い素地や⼀定の形状を持つ素地に適した調整⽅法。既知の厚さのフォイル2枚で、測定する乾燥塗膜を挟みます。

ゼロ

S, T

素地の形状や粗さがわからない場合や、素地が⼿に⼊らない場合に、測定値を補正する⽅法。ユーザーが指定した補正値が、読み取り値に適⽤されます。

S, T

繰り返し検査する場合に最も適した⽅法。
まず、既知の校正⽤フォイルの厚さを膜厚計に保存しておきます。膜厚計を調整するようにという指⽰が表⽰され、読み取り値が保存済みのフォイルの厚さに従って⾃動的に調整されます。短時間で簡単に調整できる⽅法です。

オフセット

⾃動

www.elcometer.com

jp-6

R

9 調整⽅法（続き）
校正⽅法

モデル

説明

ISO

S, T

ISO19840に従って、調整⽅法を［ゼロオフ
セット］に設定し、［カウントアベレー
ジ］を5に設定します。

SSPC PA2

S, T

SSPC PA2に従って、調整⽅法を粗⾯/2ポイントに設定し、［カウントアベレージ］を3に設定します。

スエーデン

S, T

スウェーデン規格に従って、調整⽅法を粗⾯/2ポイントに設定し、［カウントアベ
レージ］を5に設定します。

オーストラリア

S, T

オーストラリア規格に従って、調整⽅法を［ゼロオフセット］に設定し、［カウントアベレージ］を5に設定します。

10 測定モード（モデルT）
10.1 測定モード
Elcometer456には、スタンダード、オートリピート、スキャンの3つの測定モードがあります。測定モードを選択するには、メニュー→設定→測定モードを選択します。


オートリピートモードとスキャンモードは、Elcometer 456⽤ウルトラスキャン
プローブを使⽤する場合のみ選択できます。詳しくは、Elcometerに問い合わせ
るか、www.elcometer.comをご覧ください。

10.2NAVSEA モード
読取り値を10mils(254µm)以上では1mil(1µm)単位で、10mils(254µm)未満では0.1mil(0.1µm)単位で表⽰します。
NAVSEAモードを選択するには、メニュー→設定→NAVSEA、またはバッチ→新しいバッチ→NAVSEAを押します。

jp-7

www.elcometer.com

R

11 読み取り値のバッチ保存（モデルSおよびT）
1
2
3
4
5



バッチソフトキーを押して、読み取り値のバッチ保存⽤メモリを使⽤できるようにします。
読み取り値を追加するには、［新規のバッチ］または、［既存のバッチを開く］を選択します。
バッチのデータをコピーまたは確認します。
バッチ名の変更、バッチの読み取り値の消去、またはバッチの削除を⾏うには、［バッチの編集］を選択します。
⼀定の数の読み取り値をバッチに保存するには、［バッチサイズの固定］を選択します。 保存した読み取り値が、指定した数に達すると、新しいバッチが⾃動的に作成され、元のバッチと連結されます。たとえば、「NewBatch_1」というバッチは、
NewBatch_2、NewBatch_3（以下同様）になります。
それぞれの読み取り値をメモリに保存することも、［カウントアベレージ］を選択して、あらかじめ設定した数の読み取り値の平均を保存することもできます。

12 グラフの表⽰
12.1棒グラフ
読み取り値の最⼤値と最⼩値、平均値と共に、膜厚値を棒で⽰します。棒グラフを表⽰するには：1

ディスプレイソフトキーを押し、［読取値を棒グラフ］を選択します。

調整 バッチﾃﾞｨｽﾌﾟﾚｲ

メニュー

読み取り値が、設定した範囲内に収まっていない場合は、⽩い棒と読み取り値が⾚になります。
スキャンモード（モデルT）で測定しているときは、棒グラフが⾃動的に表⽰されます。

12.2ランチャート（モデルSおよびT）
最後の20個の読み取り値の推移を⽰すラン
チャートを表⽰するには：
1
2
3


バッチソフトキーを押します。
［新規のバッチ］または［既存のバッチを開く］を選択します。
ディスプレイソフトキーを押し、［読取値とランチャート］を選択します。

調整 バッチﾃﾞｨｽﾌﾟﾚｲ

メニュー

バッチの限度値を設定している場合は、範囲外の値が⾚い点で⽰されます。

www.elcometer.com

jp-8

R

12 グラフの表⽰（続き）
12.3バッチデータのグラフ（モデルT）
バッチデータのグラフを表⽰するには：
1
2



jp-9

バッチ→バッチの再検討を押し、⽬的のバッチの名前を選択します。
［バッチグラフ］を選択します
バッチの限度値を設定している場合は、範囲外の値が⾚い棒で⽰されます。
読み取り値を1つ1つ⾒ていくには、ズームソフトキーの次に  または  ソフ
トキーを押します。

www.elcometer.com

www.elcometer.com

統計の選択
全てのPSPC値
読取数
平均
最小読取値
最大読取値
標準偏差
変動係数
下限値
限度より下の数
上限値
限度より上の数
Elcometerｲﾝﾃﾞｯｸｽ値
NDFT値
NDFTより上か同じ
90-100% NDFT
IMO90：10合格/不合格

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

nは=1,2または3

調整メモリー n
保存 hh:mm dd/mm/yy
選択
調整する
調整方法
調整テスト
名前変更調整メモリー n
調整データを見る
消去調整メモリーn

調整方法
ゼロ
平滑面
粗面／２ポイントゼロ オフセット
自動統計の消去
選択された統計を見る
全ての統計を見る
統計の選択
カウントアベレージ

統計

調整
調整する
調整方法
調整メモリー1
調整メモリー2
調整メモリー3
工場調整
調整固定
調整テスト
調整チェックバッチ
空きメモリー
新規のバッチ
既存のバッチを開く
バッチングから出ますか？
バッチの編集
バッチをコピーする
バッチのレビュー削除された読取値

統計の消去
読取値のみ
読取値と選択済み統計
読取値とランチャート
読取値と棒グラフ
バッチグラフ統計

ﾃﾞｨｽﾌﾟﾚｲ

ﾒﾆｭｰ
最終読取値の削除
設定
限度設定
Bluetooth
機器情報
リセット
緊急ライト

○

○

○
○
○
○

○

○

○

削除された読取値
タグと共に削除
タグなしで削除

統計
バッチ情報
調整情報
読取値
バッチグラフバッチのレビュー
バッチ 1

バッチの編集
バッチの名前の変更
バッチの消去バッチの削除新規のバッチバッチ 開ける 1
バッチの名前の変更 1
バッチ調整
バッチタイプバッチ限度
バッチサイズの固定
NAVSEA

機器情報
プローブ情報
連絡先
法律

機器情報

○
○

○

BluetoothBluetooth有効○
Bluetooth ID
xxxxxxBluetoothコード
xxxxxx
AI

限度設定
限度を有効
上限を設定
下限を設定
NDFTを設定

設定
ユニット
言語
時間／日付
警告音のボリューム
画面の設定
プローブ
測定モード
計測器⾃動OFF調整固定
NAVSEA

○

○
○
○
○
○
○

時間／⽇付
時間設定
日付設定
フォーマットをセット
日時表示
画面の設定
画面の明度
スクリーンタイムアウト
⾃動表⽰ﾛｰﾃｰｼｮﾝ
オープニングスクリーン
プローブ
⾃動FNF
磁性F
⾮磁性 N
プローブスイッチオン

バッチ限度
限度を有効
上限を設定
下限を設定
NDFTを設定

バッチタイプ
標準
カウントアベレージ
IMO PSPC

バッチ調整
調整する
調整方法
使⽤ 調整メモリー1
使⽤ 調整メモリー2
使⽤ 調整メモリー3

特許
法律上の表示
規制

法律

○

○
○
○

○
○
○

○
○
○
○

測定モード
スタンダードモード
オートリピートモード
スキャンモード
ﾎｰﾙﾄﾞ付きｽｷｬﾝﾓｰﾄﾞ
調整固定
PINを有効にする

○
○

磁性 F1
磁性 F2

プローブ

○
○
○
○

ユニット
µm
mm
mils
inch

規制

調整⽅法
ゼロ
平滑面
粗面／２ポイントゼロ オフセット
自動ISO
SSPC PA2
スエーデン
オーストラリア

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

プローブ
接続されているプローブによって、メニューの画⾯の表⽰が変
わります。

R

13 メニュー構成-モデルTjp-10

jp-11

統計の選択
全てのPSPC値
読取数
平均
最小読取値
最大読取値
標準偏差
変動係数
下限値
限度より下の数
上限値
限度より上の数
Elcometerｲﾝﾃﾞｯｸｽ値
NDFT値
NDFTより上か同じ
90-100% NDFT
IMO90：10合格/不合格

○
○
○
○
○
○
○
○
○
○
○
○
○
○
○
○

調整方法
ゼロ
平滑面
粗面／２ポイントゼロ オフセット
自動

統計
統計の消去
選択された統計を見る
全ての統計を見る
統計の選択
カウントアベレージ

○
○
○
○
○
調整する
調整方法
工場調整
調整固定
調整テスト

調整

バッチ
空きメモリー
新規のバッチ
既存のバッチを開く
バッチングから出ますか？
バッチの編集
バッチのレビュー削除された読取値

統計の消去
読取値のみ
読取値と選択済み統計
読取値とランチャート
読取値と棒グラフ
統計

ﾃﾞｨｽﾌﾟﾚｲ

ﾒﾆｭｰ
最終読取値の削除
設定
限度設定
Bluetooth
機器情報
リセット
緊急ライト

○

○
○
○
○

○

○

削除された読取値
タグと共に削除
タグなしで削除

統計
バッチ情報
調整情報
読取値

バッチのレビュー
バッチ 1

バッチの編集バッチの消去バッチの削除新規のバッチバッチ 開ける
バッチ調整
バッチタイプ

機器情報
機器情報
プローブ情報連絡先
法律○○BluetoothBluetooth有効○
Bluetooth ID
xxxxxxBluetoothコード
xxxxxx
AI

限度設定
限度を有効
上限を設定
下限を設定
NDFTを設定

ユニット
言語
時間／日付
警告音のボリューム
画面の設定
プローブ
計測器⾃動OFF
調整固定

設定

○
○

画面の設定
画⾯の明度
スクリーンタイムアウト
⾃動表⽰ﾛｰﾃｰｼｮﾝ
オープニングスクリーン

プローブ

バッチ調整

法律

バッチタイプ
標準
カウントアベレージ
IMO PSPC

調整する
調整方法

特許
法律上の表示規制

調整固定
PINを有効にする

磁性 F1
磁性 F2

○
○
○

○
○

○
○
○
○

○

時間／日付
時間設定
日付設定
フォーマットをセット
日時表示

プローブ
⾃動FNF
磁性F
⾮磁性 N
プローブスイッチオン

○
○
○
○

ユニット
µm
mm
mils
inch

規制

調整方法
ゼロ
平滑面
粗面／２ポイントゼロ オフセット
自動ISO
SSPC PA2
スエーデン
オーストラリア

QDID

○
○
○
○
○
○
○
○
○

xxxxx
xxxxx
xxxxx
xxxxx

プローブ
接続されているプローブによって、メニューの画⾯の表⽰が変
わります。

R

14 メニュー構成-モデルS

www.elcometer.com

www.elcometer.com

読取数
平均
最小読取値
最大読取値
標準偏差
変動係数
Elcometerｲﾝﾃﾞｯｸｽ値

統計の選択

○
○
○
○
○
○
○

ゼロ
平滑面
粗面／２ポイント

調整方法

統計
統計の消去
選択された統計を見る
全ての統計を見る
統計の選択
カウントアベレージ○○○

ﾃﾞｨｽﾌﾟﾚｲ

調整する
調整方法
工場調整
調整固定
調整テスト

Escape調整

削除

51.5µmデータ
最終５個の読取値
50.5 µm52.3 µm51.0 µm51.8 µm

統計の消去
読取値のみ
読取値と選択済み統計
読取値と棒グラフ
統計

機器情報
リセット
緊急ライト

設定

ﾒﾆｭｰ
最終読取値の削除

○

Ok

○
○
○
機器情報
プローブ情報
連絡先
法律

機器情報

設定
ユニット
言語
警告音のボリューム
画面の設定
プローブ
計測器自動OFF
調整固定

○

特許
法律上の表示規制

法律

調整固定
PINを有効にする

○
○

○
○
○
○

プローブ
⾃動FNF
磁性F
⾮磁性 N
プローブスイッチオン
プローブ

○
○

画面の設定
画面の明度
スクリーンタイムアウト
⾃動表⽰ﾛｰﾃｰｼｮﾝ
オープニングスクリーン

磁性 F1
磁性 F2

○
○
○
○

ユニット
µm
mm
mils
inch

規制
xxxxx

プローブ
接続されているプローブによって、メニューの画⾯の表⽰が変
わります。

R

15 メニュー構成-モデルB

jp-12

R

16 ソフトウェアの使⽤とファームウェアのアップグレード
16.1 ElcoMaster®の使⽤
ElcoMaster® を使⽤します。このソフトウェアは、Elcometer 456モデ
ルSとTに付属しています。また、elcometer.comから無料でダウンロードすることもできます。どのモデルでも、読み取り値をPCに転送して、保存したりレポートを作成したりできます。データを転送す
るには、USB接続またはBluetooth®（モデル S&T) を使⽤します。
ElcoMaster® について詳しくは、www.elcometer.comをご覧ください。
16.2ElcoMaster®モバイルアプリの使⽤（モデルSおよびT）
ElcoMaster®モバイルアプリは、検査現場での使⽤に最適です。
Android™⽤とiOS⽤があり、次の機能が搭載されています。






厚さ計で読み取った値を直接モバイル
デバイスに送信して、GPSの値と共にバッチとして保存する。
試験⾯の写真を追加する。
読み取り値を地図や写真、図表に関連付ける。
分析や報告書作成⽤に、モバイルデバ
イスにあるデータをPCに転送する。

ElcoMaster®モバイルアプリについて詳しく
は、www.elcometer.comをご覧ください。

Android 2.1以上のスマートフ
ォンとタブレットに対応した
モバイルアプリです。インストールするには、
www.elcometer.comから、またはGoogle Play™ Storeアプ
リを使ってダウンロードし、画⾯に表⽰される指⽰に従い
ます。
jp-13

www.elcometer.com

R

16 ソフトウェアの使⽤とファームウェアのアップグレード（続き）
iPhone 6 Plus、iPhone 6、iPhone
5s、iPhone 5c、iPhone 5、iPhone
4s、iPhone 4、iPad Air 2、iPadmini 3、iPad Air、iPad mini
2、iPad（第3、4世代）、iPadmini、iPad 2、iPod touch（第4、5世代）に対応。インストール
するには、
www.elcometer.com、またはAppStoreからダウンロードして、画⾯に表⽰される指⽰に従います。
16.3 ファームウェアのアップグレード
ElcoMaster®を使って、厚さ計のファームウェアを最新バージョンに
アップグレードすることができます。インターネットに接続している
PCでElcoMaster®を実⾏し、厚さ計を接続すると、新バージョンがリリースされているかどうかがわかります。
17 仕様
電源

AA電池2本、充電池も可

使⽤温度-10～50°C（14～122°F）

相対湿度

0～95%

本体⼨法

14.1 x 7.30 x 3.70cm（5.55 x 2.87 x 1.46インチ）

本体重量
電池も含む

⼀体型:156g（5.5オンス）
セパレート型:161g（5.68オンス）

適合規格：
AS2331.1.4、AS3894.3-B、AS/NZS1580.108.1、ASTMB499、ASTMD
1186-B、ASTMD1400、ASTMD7091、ASTME376、ASTMG12、BS
3900-C5-6B、BS3900-C5-6A、BS5411-11、BS5411-3、BS5599、
DIN50981、DIN50984、ECCAT1、EN13523-1、IMOMSC.215(82)、
IMOMSC.244(83)、ISO1461、ISO19840、ISO2063、ISO2178、ISO2360、ISO2808-6A、ISO2808-6B、ISO2808-7C、ISO2808-7D、ISO
2808-12、NFT30-124、SS184159、SSPCPA2、⽶国海軍PPI63101-000、
⽶国海軍NSI009-32

www.elcometer.com

jp-14

R

18 関連する法律と規制について
適合宣⾔：
Elcometer 456モデルBは以下のEU指令に適合しています：
2014/30/EU
電磁両⽴性
2011/65/EU
電気·電⼦機器における特定有害物質の使⽤制限
Elcometer 456モデルS及びTは以下のEU指令に適合しています：
2014/53/EU
ラジオ·通信端末設備
2011/65/EU
電気·電⼦機器における特定有害物質の使⽤制限
適合宣⾔は以下のリンクよりダウンロードできます：
モデルB: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/Japanese/DoC_456C_B.pdf
モデルSおよびT: www.elcometer.com/images/stories/PDFs/Datasheets/Declaration_of_Conformity/Japanese/DoC_456C_S&T.pdf
使⽤する周波数帯：2,402 - 2,480 MHz
最⼤送信出⼒：<4 dBm
本製品は、CISPR 11規格のグループ1、クラスBのISM装置に当てはまります。
クラスBに分類される装置：家庭での使⽤、および住宅⽤の低電圧配線網に直接接続される施設での使⽤に適しています。
グループ1のISM装置：装置内部の機能で必要とする無線周波エネルギーを意図的に⽣成したり使⽤したりします。
USB接続は、データ転送⽤のみに使⽤し、USB電源アダプタを使ってコンセントに接続しないでください。
本装置は、FCC規制の第15部に準拠しています。本装置は、次の2つの条件の元で使⽤するものとします。（1）本装置が⼲渉を引き起こさない。（2）本装置の望ましくない動作の原因となる⼲渉も含み、どのような⼲渉も受け⼊れる。
ACMA準拠マークは、メニュー/機器情報/法律/規制を選択すると確認できます。
Elcometer 456モデルSおよびT：技適マークとその証明番号、FCC ID、Bluetooth SIGのQDIDは、ﾒﾆｭｰ→機器情報→法律→規制を選択すると表⽰されます。
注：本装置は、FCC規制の第15部に従って検査され、クラスB、デジタル装置の限度値を満たしていることが確認されています。これらの限度値は、装置の家庭での使⽤による有害な⼲渉を妥当な範囲に抑えるために設定されています。本装置は、電磁波を⽣成、使⽤し、外部に放射します。そのため、取扱説明書どおりに設置して使⽤しないと、無線通信障害を引き起こす可能性があります。ただし、ある決まった⽅法で設置すると⼲渉が発⽣しないという保証はありません。本装置が原因で、ラジオやテレビの受信障害が発⽣していると思われる場合は、本装置の電源を⼊れたり切ったりして確かめてください。本装置が受信障害を引き起こしている場合は、次のことを試してください。
--アンテナの位置や向きを変えます。
--ラジオやテレビから離れた場所に本装置を設置します。
--ラジオやテレビを接続している電気回路（コンセント）とは別の回路に本装置を接続します。
--販売代理店または電気通信技術者に相談します。
携帯機器や基地局による無線周波数（RF）電磁波の放射に関するFCCの規制によって、このような装置の運転中はアンテナを周囲の⼈から20cm以上離さなければならないと定められています。必ず、この規制に従ってください。本装置⽤のアンテナを他のアンテナや送信機と同じ場所に設置したり、同時に使⽤したりしないでください。
Elcometer Limitedによって明⽰的に認められていない改変を本装置に加えると、FCC規制に従って本装置を操作する権利を失うことがあります。
本装置は、Industry Canada（カナダ産業省）ライセンス免除技術基準（RSS）に準拠しています。本装置は、次の2つの条件の元で使⽤するものとします。（1）本装置が⼲渉を引き起こさない。（2）本装置の望ましくない動作の原因となる⼲渉も含み、どのような⼲渉も受け⼊れる。
クラスBのデジタル装置に分類され、CAN ICES-3(B)/NMB-3(B)に準拠しています。
とElcoMaster®は、Elcometer Limitedの登録商標です。所在地：Edge Lane, Manchester, M43 6BU
United Kingdom
は、Bluetooth SIG Incが所有する商標です。Elcometer Limitedにライセンス付与されています。
Elcometer 456モデルSおよびT：iPhone 6 Plus、iPhone 6、iPhone 5s、iPhone 5c、iPhone 5、iPhone 4s、iPhone 4、iPad
Air 2、iPad mini 3、iPad Air、iPad mini 2、iPad（第3、4世代）、iPad mini、iPad 2、iPod touch（第4、5世代) に対応。
上記の「対応」とはの該当するモデルをiPod、iPhone、またはiPadに接続するためのアクセサリーがあり、Appleによる性能基準を満たしていることが開発者によって承認されているという意味です。Appleは、本製品の動作、および本製品が安全基準や規制に準拠しているかどうかについて⼀切責任を負いません。iPod、iPhone、またはiPadと上述のアクセサリーの使⽤によって、無線通信が影響を受けることがあります。
iPad、iPhone、iPod touchは、⽶国および他の国におけるApple Inc.の登録商標です。
App Storeは、⽶国および他の国におけるApple Inc.の登録商標です。
Google Playは、Google Inc.の商標です。
その他の商標については、その旨が記されています。
R

Elcometer 456は、段ボール箱に梱包されています。包材は、環境保全に配慮した⽅法で破棄してください。詳しくは、地⽅⾃治体等の適切な機関にお問い合わせください。
本社：Elcometer Limited, Edge Lane, Manchester, M43 6BU, United Kingdom.

TMA-0488-12 第17 - 表紙と本⽂ 22373-12