Genius 1024
User Manual

Genius1024 User Manual Rev 8 PN UM-0018 RSID 1917 19/03/24

Contents
Change History
How to use this Manual
Introduction
Warranties and Liabilities
Safety Notices
Symbols
Safety Notice to Users
EU Declaration of Conformity
UK Declaration of Conformity
WEEE Compliance Statement
CSA Compliance Statement
Technical Specification
Genius 1024
UnpackingFittings Kit Contents
Installation
Generator Environment
Generator Overview
General Dimensions
Rear Connections
Unit Controls
Drain Connection
Electrical Connection
Start-Up Sequence
Connecting to the application
Tubing Lengths
Normal Operation
On Demand Gas
Generator Cycling
Unusual Operation
Service Requirements
Service Schedule
Service Indication
Stage 1
Stage 2Service Indication Reset
Peak Protected
Cleaning
Run Time Alarm
Troubleshooting

Page 2

3
3
4
5
6
6
6
7
8
9
10
11
11
12
13
14
14
15
15
16
16
17
18
19
20
20
21
21
21
21
22
22
23
23
23
23
24
25
26
27

Change History
Rev

Comment

Name

1

Initial Release

L. CouttieDate
19/06/2017

2

Updates to Incorporate CSA feedback

L. Couttie

07/10/2019

3
4
5

Updates to Incorporate CSA feedback
Heat Output added to Tech Spec

L. Couttie
L. Couttie

21/10/2019

Declarations Update

L. Couttie

15/11/2021

6

Power Cable Update

L. Couttie

01/08/2022

7

Drain Container Warning Added

L. Couttie

28/10/2022

8Spec. Update

L. Couttie

19/03/2024

09/03/2019

How to use this ManualThis manual is intended for end users and has been written as a reference document
where you can skip to the relevant information.
Users can refer to the contents page to find the relevant information.
Please review each of the following sections carefully.
Thank you for selecting Peak Scientific to meet your gas generation needs, and should
you require any further assistance or support please do not hesitate to contact Peak
Scientific or the Peak Partner from which you purchased your generator.

Page 3

Introduction
The Genius 1024 has been developed to cater for the requirements of the Sciex range of
LCMS instruments.
This model provides a source of Nitrogen gas as well as two independent supplies ofclean, dry air; other features include:•

Small in size – fits under a standard lab bench•Anti- vibration – maximum reduction of vibration•

Service indication – allowing you to plan your maintenance and keep your LCMS
uptime at a maximum•

Improved drainage – reduction of moisture carry over and thus increased 		reliability•

Re- heat technology – improves membrane performance and reliability•

Robust control system – improves safety and reliability of units

With the Genius 1024 based on proven technology, it selectively removes oxygen,moisture and other gases to leave clean, dry, phthalate free Nitrogen and Dry Air. Twointernal air compressors make this unit independent from in- house air supplies andfitted castors allow the user to easily position the unit in the lab.
To ensure this Generator model meets our high expectations with regards to reliability
and performance, we have tested this new model extensively at our manufacturing
plant and with end users around the world to ensure reliability and longevity of thesystem.

Page 4

Warranties and Liabilities
1. The Company warrants that it has title to the Goods.
2. Subject to the provisions of this clause the Company warrants that the Goods 		
shall comply in all material respects with any specification referred to in the Order
Confirmation (as the same may be amended) and shall, subject thereto, be free 		from defects in material and workmanship for the lesser of a period of twelve 		months from the date of delivery or thirteen months from the date of dispatch 		from the factory.
3. Save as provided in this clause and except where the Goods are sold to a person 		
dealing as a consumer (within the meaning of the Unfair Contract Terms Act 		1977) all warranties, conditions or other terms implied by statute or common
law are hereby expressly excluded save to the extent they may not be lawfully
excluded. When the Goods are sold to a consumer within the meaning of the Unfair
Contract Terms Act 1977 their statutory rights are not affected by the provisions of
this clause.
4. In the event of the Customer making a claim in respect of any defect in terms of 		clause 2 hereof the Customer must.
1. Reasonably satisfy the Company that the Goods have been properly installed,commissioned, stored, serviced and used and without prejudice to the generalityof the foregoing that any defect is not the direct or indirect result of lack of repair
and/or servicing, incorrect repair and/or servicing, use of wrong materials and/or
incorrect spare parts
2. Allow the company to inspect the Goods and/or any installation and any relevant
packaging as and when reasonably required by the Company.
5. Subject to the Company being notified of any defect as is referred to in sub-		
clause 2 hereof within a reasonable time of it becoming apparent and subject always
to the terms of sub-clause 4 hereof, the Company shall, in its option, replace orrepair the defective Goods or refund a proportionate part of the Price. The Company
shall have no further liability to the Customer (save as mentioned in sub-clause 6
hereof).
6. The Company shall be liable to indemnify the Customer in respect of any claim
for death or personal injury to any person in so far as such is attributable to thenegligence or breach of duty of the Company or any failure by the Company tocomply with the provisions of sub-clause 2 hereof.
7. Save as provided in sub-clause 2 hereof the Company shall not be liable in respectof any claim by the Customer for costs, damages, loss or expenses (whether direct,
indirect, consequential or otherwise) or indemnity in any respect howsoever arising
including, but not by way of limitation, liability arising in negligence (other thanpursuant to clause 6 above) that may be suffered by the Customer or any third
party.

Page 5

Safety Notices
Peak Scientific Instruments cannot anticipate every possible circumstance which may
represent a potential hazard. The warnings detailed within this manual refer to the most
likely potential hazards, but by definition cannot be all inclusive. If the user employs
an operating procedure, item of equipment or a method of working which is not
specifically recommended by Peak Scientific, the user must ensure that the equipment
will not be damaged or become hazardous to persons or property.

Symbols
This manual uses the following symbols to highlight specific areas important to the safeand proper use of the generator.

WARNINGA WARNING notice denotes a hazard. It calls attention to an operating procedure,
process or similar, which if not correctly performed or adhered to, could cause
personal injury or in the worst case death. Do not proceed beyond a WARNING
notice until the indicated conditions are fully understood or met.

CAUTIONA CAUTION notice denotes a hazard. It calls attention to an operating procedure,
process or similar, which if not correctly performed or adhered to, could cause
damage to the generator or the application. Do not proceed beyond a CAUTION
notice until the indicated conditions are fully understood or met.
Caution, risk of electric shock. Ensure power to the generator has been removed
before proceeding.

Safety Notice to Users

WARNINGWARNING

These instructions must be read thoroughly and understood before
installation and operation of your Peak Generator. Use of the generator
in a manner not specified by Peak Scientific MAY impair the SAFETY
provided by the equipment.
When handling, operating or carrying out any maintenance, personnel
must employ safe engineering practices and observe all relevant local
health and safety requirements and regulations. The attention of UK
users is drawn to the Health and Safety at Work Act 1974, and theInstitute of Electrical Engineers regulations.
If the equipment is used in a manner not specified by the manufacturer,
the protection provided by the equipment maybe impaired.

WARNING

Page 6

EU Declaration of Conformity
We			

Peak Scientific Instruments Ltd.

Of			

Fountain Crescent, Inchinnan, Renfrewshire, PA4 9RE

Hereby declare that, this declaration of conformity is issued under the sole
responsibility of the manufacturer.
Equipment Type:		

Nitrogen Gas Generator

Model Designator:

Genius 1024

To which this declaration relates, is in conformity with the following applicable EU
Directives, harmonized standards, and other normative requirements.
•

Low Voltage Directive 2014/35/EU
EN 61010-1: 2010 Safety Requirements for Electrical Equipment for Measurement,
Control and Laboratory Use.

•

Electromagnetic Compatibility Directive 2014/30/EU
EN 61326-1: 2013 Electrical Equipment for Measurement, Control and Laboratory Use– EMC Requirements. (Class A)•

Restriction on the use of certain hazardous substances in electronic equipment(RoHS) Directive 2011/65/EU as amended by EU 2015/863.

•

FCC 47 CFR Part 15 class A
Unintentional radiators; Conducted and Radiated emissions limits.

Signed for and on behalf of Peak Scientific by		
Signed:				
Name:

Fraser Dunn

Position:

Design Engineering Manager

		Peak Scientific Instruments ltd,		Inchinnan, Renfrew, Scotland, PA4 9RE, UK.

Date:		

10th August 2021

Page 7

UK Declaration of Conformity
We			

Peak Scientific Instruments Ltd.

Of			

Fountain Crescent, Inchinnan, Renfrewshire, PA4 9RE

Hereby declare that, this declaration of conformity is issued under the sole
responsibility of the manufacturer.
Equipment Type:		

Nitrogen Gas Generator

Model Designator:

Genius 1024

To which this declaration relates, is in conformity with the following applicable UK
Statutory Instruments, Standards and other normative requirements.
•

The Electrical Equipment (Safety) Regulations 2016 (SI 2016 / 1101) as amended.
BS61010-1:2010 Safety Requirements for Electrical Equipment for Measurement
Control and Laboratory Use.

•

The Electromagnetic Compatibility Regulations 2016 (SI 2016 / 1091) as amended.
BS61326-1:2013 Electrical Equipment for Measurement , Control and Laboratory Use– EMC Requirements.

•

The Restriction of the Use of Certain Hazardous Substances in Electrical andElectronic Equipment Regulations 2012 (SI 2012 / 3032) as amended.

Signed for and on behalf of Peak Scientific by		
Signed:				
Name:

Fraser Dunn

Position:

Design Engineering Manager

		Peak Scientific Instruments ltd,		Inchinnan, Renfrew, Scotland, PA4 9RE, UK.

Date:		

10th August 2021

Page 8WEEE Compliance Statement
The Waste Electrical and Electronic Equipment (WEEE) Regulations SI 2013 No 3113
and or the Waste Electrical and Electronic Equipment (WEEE) Directive 2012/19/EU
apply to all electrical and electronic equipment placed on the market in the UK and EU
covered by the scope of regulations which can be found in the Government GuidanceNotes (PDF) produced by the Department for Business Innovation and skills for the UK
and here for Europe.
All PEAK products that are subject to the WEEE directive are compliant with the
WEEE marking requirement. Such products are marked with the “crossed-out wheeliebin” symbol (shown below) in accordance with European standard EN50419. All
old electrical equipment can be recycled. Please do not dispose of any electricalequipment (including those marked with this symbol) in general rubbish bins. Pleasecontact your dealer or distributor for clarity.

Page 9

CSA Compliance Statement
CSA Group (Canadian Standards Authority) is a Nationally Recognised Testing
Laboratory (NRTL), headquartered in Toronto Canada.
They are authorised to evaluate product to both their own and Underwriters Laboratory(UL) standards and certify the product to be in compliance to the relevant standards.
Peak products are certified to the current in force revision of the following standards
in order to cover both Canadian and United States requirements for “Safety
Requirements for Electrical Equipment for Measurement, Control and Laboratory use,
Part 1: general Requirements”.
Canada:CAN/CSA C22.2 No 61010-1-12

United States:

UL 61010-1

As a result the products covered by this statement are certified and listed by CSA
accordingly and are entitled to carry the CSA mark with both Canadian and UnitedStates subscripts , as shown below on the product rating label.

Page 10

Technical Specification
Genius 1024
Environment
Minimum Operating Ambient Temperature
Maximum Operating Ambient Temperature
Maximum Altitude
Maximum Relative Humidity
Minimum Storage Temperature*
Maximum Storage Temperature*

5°C (41°F)
30°C (86°F)
3000 m
80% Non-Condensing-20°C (-4°F)
60°C (140°F)*NOTE – When taken out of storage the Generator should be allowed to acclimatize at room temperature
for a minimum of 3 hours before operation.

Generator Outlets
Curtain Maximum Flow^
Source Maximum Flow^
Exhaust Maximum Flow^
Particles
Phthalates
Suspended Liquids
Gas Outlets
Drain Outlet
Pressure Gauges

19 L/min @ 4.48 bar (0.670 cfm @ 65 psi)
26 L/min @ 6.89 bar (0.918 cfm @ 100 psi)
25 L/min @ 4.13 bar (0.882 cfm @ 60 psi)
<0.01µm
NONE
NONE
3 x ¼” BSPP
1 x ¼” BSPP
3

^NOTE – These performance characteristics are valid only when paired with an approved Sciex application.
Please refer to current application matrix for a list of supported products.
Electrical Requirements
VoltageFrequency
Current
Input Connection
Power Cord
Pollution Degree
Insulation Category220-240V ± 10%50/60 Hz
7 AmpsC20 PlugC19 socket to local connection
2
II

GeneralDimensions cm (inches) WxDxH
Generator Weight Kg (lbs)
Shipping Weight Kg (lbs)
Heat Output

Page 1160 x 75 x 71.2 (23.6 x 29.5 x 28)
108.5 (239)134 (295)
3775 BTU/hr

Unpacking
Although Peak Scientific takes every precaution with safe transit and packaging, it isadvisable to fully inspect the unit for any sign of transit damage.
Check ‘SHOCKWATCH’ and ‘TIP-N-TELL’ labels for signs of rough handling prior to
unpacking.

Any damage should be reported immediately to the carrier and Peak Scientific or thePeak Partner from where the unit was purchased.
Follow the unpacking instructions posted on the side of the crate. It will require two
people to remove the unit from the shipping crate and to manoeuvre the generator to
the desired location.
Please save the product packaging for storage or future shipment of the generator.
Note: Included with the generator is a “Fittings Kit” containing mains power leads for
UK, EU & US and also all the required fittings and warranty registration card. Be careful
not to discard these with the packaging.

Page 12Fittings Kit Contents
Supplied in the Fittings Kit are all the fittings required to connect the generator to theapplication. The contents of the Fittings Kit are as follows:
1. ¼” Teflon Tubing							x 9m

2. 6mm PE Tubing							x 3m
3. ¼” Compression Fitting						x 3

4. 6mm Push-Fit Fitting

				x 1

5. Flow Control Silencer						x 36. UK Mains Power Cable 230v					

x17. EU Mains Power Cable 230v					

x18. US Mains Power Cable 230v					

x19. 8mm Hex Key								x 1All of the generators output ports are located on the output panel at the rear of theunit.

Page 13

Installation
Generator Environment
The generator should be situated in a well ventilated environment, with at least a
75mm (3”) air gap all around it and should be positioned such that it can be easily
disconnected from the mains supply if required. If this is not convenient then the unit
can be sited elsewhere, however, consideration should be made of the lengths of piperuns as pressure drops can result from extended runs of pipe.
Performance of the generator (like all sophisticated equipment) is affected by ambientconditions. Note should also be taken to the proximity of Air Conditioning outlets.
These can sometimes give rise to “pockets” of air with high relative humidity. Operationof the unit within such a pocket could adversely affect its performance. Consideration
should also be given to the air flow around the unit. It is recommended that an air gapof 75mm (3”) should be maintained down both sides and at the rear of the unit. Pleaserefer to the drawing on the following page for the general dimensions of the unit.
Minimum Operating Ambient Temperature:

5 °C (41 °F)

Maximum Operating Ambient Temperature:

30 °C (86 °F)

Page 14

Generator Overview
General Dimensions
600 mm / 23.6”

750 mm / 29.5”712 mm / 28"
The generator must always be placed on a flat, level surface. Failure to
do so will affect the performance of the generator.
WARNING

Note: Gauges are for indication purposes only.

Page 15

Rear Connections
6mm Push-Fit Fitting¼" Compression Fitting

Mains Cable

CURTAIN

Unit Controls

Pressure Gauges
Low Pressure LED
Service LED
Mute Button
Power Switch

Note: Gauges are for indication purposes only.

Page 16

SOURCEEXHAUST

Drain Connection
Fit the 6mm push fit fitting to the drain port located on the output panel. Tighten
using a 16mm or 5/8” spanner. Use the 6mm tubing to connect this to a suitable drain
connection or container. It should be noted that the Generator can expel a considerable
amount of water from this (dependant on ambient humidity).

CAUTION

Page 17

If a container is used it should be emptied at regular intervals. The
container used MUST be a plastic material and MUST NOT be glass
for safety reasons. The container MUST NOT have an airtight seal aswater and air will be expelled periodically under slight pressure.

Electrical Connection
Connect the generator to an appropriate 230 volt single-phase supply, refer to thegenerator serial plate for input specification and ensure your supply matches therequirements.
If an appropriate mains power cords is not supplied or a substitute one is used thenensure that all components of it the plug, cord and connector have adequate ratings forthe generator and appropriate approvals for the country of use.

WARNING

This unit is classified as SAFETY CLASS 1. THIS UNIT MUST BE
EARTHED. Before connecting the unit to the mains supply, pleasecheck the information on the serial plate. The mains supply must be of
the stated AC voltage and frequency.

EARTH/GROUND (E):-		

Green & Yellow

or

Green

LIVE (L):-				

Brown

or

Black

Neutral (N):-			

Blue

or

White

		

Electrical requirements are 230VAC nominal +/- 10%. However, running continuously at
voltages outwith this is not recommended. Extended periods at extremes can have a
detrimental effect on the operation and life of the generator.
If the equipment is used in a manner not specified by the manufacturer,
the protection provided by the equipment maybe impaired.
WARNINGTo ensure that the correct voltage is being supplied to the generator, the generator
comes equipped with an inbuilt voltmeter. This is located on the underside of thecompressor compartment, and will measure and display the mains voltage that is being
supplied to the generator.
The voltmeter should be checked prior to the initial purge of the system.
If the reading is 219V or less, then we would highly recommend fitting a Dual Tap
Transformer 06-3200. This can be ordered directly from Peak Scientific.

Page 18

Start-Up Sequence

CAUTION

Before the Generator is connected to the application, the Generator
should be operated in isolation (i.e. not connected to the application)
for thirty minutes. This is to ensure any impurities present are purged
from the system. Failure to do this may harm the application.

Following the voltage check, (see Page 16), the unit should be switched off andsilencers fitted to each of the three outlet ports of the output panel (ref Install Guide).
Once this is done, the Generator can be re-connected to the mains and switched on.
Pressure will start to build in the internal storage tanks which can be monitored bywatching the output pressure gauge on the front panel. This will climb to the factory setpressure as noted in the specifications.
Once this pressure is reached, the compressors will continue to run until the internaltank upper pressure limit is reached and the compressor has run for a period of at least
two and a half minutes.
The compressors will then rest until the internal tank lower pressure limits is reached.
Once this limit is reached the compressors will switch back on again. This compressor
cycling is normal and will continue throughout the operation of the Generator.
When the system has been operated for a period of 30 minutes, all the internal pipework and storage tanks will have been purged with Nitrogen.
The Generator is now purged and the tubes can be connected at the rear of the unit.

Page 19

Connecting to the application
Once the initial purge run of 30 minutes has completed, and the generator has beenrunning for 1 hour, (required to reach purity for the application) it is ready to beconnected to the application(s)The pressure in the internal storage tanks must be allowed to dissipate
before connecting the generator to the application(s)CAUTION

Attach the ¼” compression fittings to the outlets of the generator. Using the ¼” tubing
supplied, connect the outlets of the generator to the inlets on the application.
If you require more tubing than is supplied please refer to the Tubing Lengths section.

CAUTION

Once the tubing is connected to the application, please ensure that itis thoroughly checked for being leak-tight. Even the slightest leak in
the gas supply between the generator and the application can lead toa reduction in efficiency.

Tubing Lengths

CAUTION

The diameter of the tubing which will be connected to the gas outlets
is important and is determined by the length of tubing required. Failure
to follow these recommendations could lead to pressure between
generator and application.

< 10 meters:		

Use ¼”/3⁄16” (¼” O/D, 3⁄16” I/D) P.T.F.E. tubing.

> 10 - 40 meters: Use 3⁄8”/ 5⁄16” (3⁄8” O/D, 5⁄16” I/D). Tubing and fittings not supplied
				in the fittings kit.
> 40 metres:				Please contact Peak Scientific with the relevant distance andwe will calculate the flow resistance and the tubing size required.

A combination of ¼”/3⁄16” and 3⁄8”/5⁄16” tubing may be used to ensure that there is no large
diameter tubing within the lab (i.e. for the first 20 meters from the generator use 3⁄8”/5⁄16”
and the final 10 meters to the application use ¼”/3⁄16” tubing). Keep the connections andbends to a minimum.

Page 20

Normal Operation
The generator is designed specifically to minimize operator involvement. Given that thesystem is installed as described in earlier sections and is serviced in accordance withthe specified maintenance recommendations (see Service Requirements), then it should
simply be a matter of turning the generator on when it is required.
The generator will automatically produce the factory set flow and pressure as detailed
in the Technical Specifications.

On Demand Gas
When the application requires a gas flow, the generator will produce gas on demand.
The generator will continue to supply gas whilst the application is in a ‘Standby’ mode.
As soon as a demand is detected from the application, the generator will match theapplications requirements.

Generator Cycling
The generator is designed to operate in either a cyclic mode or run continuously, thiswill be dependent on the output flow.
The cyclic rate of the compressors will depend on the gas demand of the application;
the generator will cycle when the upper tank pressure set limits are achieved within a
3-minute time period.
When the application demands increase, the compressor duty will be higher and therest period of the compressor cycle will be shorter. If operated at maximum flow, thegenerator will run continuously to maintain the upper tank pressures. Another cause forcontinuous operation would be if the generator is installed: at high altitude, in an extreme environment or subjected to low voltage supply. The compressors will undergo
periods where they do not cycle.

Unusual Operation
If at any time the generator begins to emit excessive noise or vibration, then it should
be switched off and you should contact Peak Scientific or the Peak Partner from which
the generator has been purchased.

Page 21

Service Requirements
Service Schedule
Purchase Interval

Component

Visit

12 Months

Genius 1024 Annual Maintenance Kit*

www.peakscientific.com/ordering* Please quote your generator serial number when ordering your Annual MaintenanceKit to ensure selection of the appropriate kit for your generator.

Page 22

Service Indication
The generator has the following Service Indication Stages:-Stage 1Once either compressor requires a service the LED indicator (yellow) on the front of thegenerator will illuminate.
This is to make the user aware that a service of the generator is due and should beplanned at the earliest convenience. The generator will continue to operate as normal
with the LED on.

Stage 2
If the service is not completed the generator will continue to run. After 2 weeks, theservice LED indicator (yellow) will start to flash.
This is to make the user aware that the service of the generator is now overdue andmust be completed immediately to ensure the continuous trouble free operation of thegenerator.

Service Indication Reset
Once the service has been completed the Service Indication LED can be reset in themain control PLC. This will be performed by the Peak Service Engineer or trained
service representative that completes the service operation.

Page 23

Peak ProtectedWith Peak Scientific you invest in not only a product but peace of mind. With a network ofcertified Peak engineers stationed throughout the globe, Peak’s rapid response team arenever far away and our commitment is to keep your generator running day in, day out,
protecting your laboratory workflow.

[Peak Protected] can provide…

Installation

A dedicated Peak engineer will visit
your lab to install and setup yourgenerator

IQ/OQ

Certified assurance for applications
requiring documented qualification

Complete plan

Swift response by a Peak Service
Engineer within 72 hours & planned
preventative maintenance

SparesGenuine Peak parts with expressdelivery, ensuring optimal
performance and lifetime

Premium Protected

Guaranteed rapid on-site response
within 24 hours & planned
preventative maintenanceTechnical Support Hotline

Around the clock support by phoneor online with our global technicalhelpdesk

To find out more about protecting your investment visit: www.peakscientific.com/protected

Page 24

Cleaning
Clean the outside of the generator only using warm soapy water and a clean damp
cloth. Ensure all excess fluid is thoroughly removed from the cloth prior to use.
Cleaning should only be undertaken with the power switched off andthe power cord removed from the rear of the generator.

CAUTION

Under no circumstances should any solvents or abrasive cleaning
solutions be used as these can contain fumes that could be harmful tothe generator.
Care should be taken with Leak Detections Liquids.

CAUTION

Page 25

Run Time Alarm
The Genius 1024 monitors the operating tank pressures, in this case the N2. When thisis at the set point, there is no alarm. Only when the pressure drops to a pre-determined
set point will the alarm be activated, indicating LOW pressure.
It should be noted that the Generator will continue to operate so long as it canmaintain pressure during these alarms.
Note: the buzzer may be temporarily silenced for approximately 90 minutes bypressing the “MUTE” button on the front facia; the Alarm LED will remain lit.

Page 26

Troubleshooting
Problem

The Generator will not switch on and the power switch
does not illuminate.

Compressors are running but pressure is not building.

Possible Solution•

Ensure power cord is plugged into the Generator
and that the power socket is turned on.

•

Check the fuse in the power cord plug.

•Contact your service provider.

•Contact your service provider.

•

Check pressure gauges are showing normalpressure.

•Contact your service provider.

•

A compressor(s) due for service. Contact yourservice provider.

•

Refer to Service Indication section of this manual forfurther information.

•

A compressor(s) has not cycled for 8 hours. Refer tothe Run Time Alarm section of this manual.

•

Ensure ambient temperatures are within thespecification.

•

Turn the Generator off and on to reset the alarm andcontact your service provider.

The mass spec is reporting low pressure.

Yellow “SERVICE” LED on front panel is on constantly.

Yellow “SERVICE” LED on front panel is flashing.

Page 27

Go Online or Complete and Return
We know that registering any of your recently purchased products is not thefirst thing on your mind- but it is very important to both of us. Not all warranties
are alike and Peak Scientific stand out against other gas suppliers as we offer acomprehensive, quick response, on-site warranty. This means that in the very
unlikely case that your gas generator develops a fault we have rapid support
teams on-hand around the world who are able to come to your lab and get you
back up and running in no time.
Register for your comprehensive 12 month on-site warranty with ease online atwww.peakscientific.com/protected.
Alternatively, you can send the completed form to Peak Scientific by post or
email at warranty@peakscientific.com.

Go Online or Complete and Return
You can register for your FREE 12 month Warranty with ease online atwww.peakscientific/protected.
Alternatively, you can send the completed form to Peak Scientific by post or
email at warranty@peakscientific.com.

Product Warranty Registration
Contact name
Email addressCompany
Address
City/town
PostcodeCountry
Telephone
Generator serial #Model typeInstallation date
Do you still use analternative gas solution i.e.
cylinders or bulk liquid?

Yes

What gas requirements do
you have in your lab?

HydrogenNo

NitrogenZero Air

Extend your cover with
Peak Scientific offer comprehensive gas generator after sales support packages.
Peak [Protected] aftercare support can guarantee an on-site response within
72 hours*, genuine parts from our ISO9001 approved factory and a 95% first-time
fix rate. See our enclosed Peak [Protected] leaflet for further information.
Important!
You have 1 month to register your Peak Scientific product from the date of installation. Once registered
the warranty will be honoured for a period of 12 months. If you wish to defer the installation of yourgenerator, you must notify Peak Scientific immediately by emailing warranty@peakscientific.com.
For generators that remain unregistered after 1 month from the shipment date, the warranty will beconsidered active from the date of factory dispatch.
* Complete Plan only






Important!
You have 1 month to register your Peak Scientific product from the date of
installation. Once registered the warranty will be honoured for a period of 12months. If you wish to defer the installation of your generator, you must notify
Peak Scientific immediately by emailing warranty@peakscientific.com. For
generators that remain unregistered after 1 month from the shipment date, thewarranty will be considered active from the date of factory dispatch.
Page 28

Page 29

Page 30

Page 31

Peak Scientific has highly trained, fully certified Field Service Engineers located in over 20 countries
across every continent around the world. This allows us to provide an industry-leading rapid response
service to our customers. With [Peak Protected], your laboratory’s productivity becomes our top
priority.
To discuss Peak Protected generator cover and payment options speak to your local Peak
Representative or for further information contact: protected@peakscientific.com

Peak Scientific
Fountain Crescent
Inchinnan Business Park
Inchinnan
PA4 9RE
Scotland, UK
Tel: +44 141 812 8100
Fax: +44 141 812 8200
For further information on any of ourgenerator products please contact
marketing@peakscientific.com
Genius1024 User Manual Rev 8 PN UM-0018 RSID 1917 19/03/24