User’s Manual
Home-care Beauty Device
Photo PLUS Prestige Pro
M30

Table of Contents
Safety Precautions • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 1
Use in Foreign Countries • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 6
Names of Components • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 7
Before Use• • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 8
How to Use • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 13
After Treatment • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 17
Troubleshooting • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 18
After-sales Service • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 21
 Product Specifications• • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • • 22

Thank you for purchasing this YA-MAN product.
 Please read this User’s Manual carefully for information on how to use the device properly andsafely.
 After reading the contents hereof, be sure to keep this User’s Manual in an accessible location.
 Be sure to include this User’s Manual if you are transferring ownership of the device to anotherperson.
M30-1-001E
23

Safety Precautions
WARNING
The following individuals should not use the device:
Failure to comply may cause accidents or other problems.
• If you have existing or suspected heart disease, or related disorders• Infants or if you are unable to express your intentions on your own• If you are unable to sense temperature or stimulation• If you have an autoimmune disease

Consult a physician prior to use under the following conditions:• If you have an acute disease • If you have an infectious disease • If you have a malignant tumor• If you have a febrile disease • If you have high blood pressure or a blood disease• If you have a fever• If you have sensory impairment due to severe peripheral circulatory dysfunction (e.g. diabetes)• If you have a nerve damage • If you have a skin disease or atopic dermatitis• If you are a hospital outpatient and are taking medication• If you have a pre-existing skin condition, such as an allergic constitution or contact dermatitis,
or if you have a concern about your skin when using this device

 Classifications of Precautions

DANGER

Improper use of this device poses a severe and likely risk of death orserious injury.

WARNING

Improper use of this device may pose a risk of death or serious injury.

ATTENTION

Improper use of this device may cause an injury or property damage*.

Safety Precautions

Safety Precautions• Read these Safety Precautions carefully before using the device in order to use it correctly.
• Use the device sensibly in accordance with the prescribed steps.
• Immediately discontinue use and arrange to have the device repaired if any abnormality occurs.
• Do not use the device while it is in a defective state, as doing so could cause physical harm to the user.
• The precautions listed in the User’s Manual are very important. For your safety, please follow themcarefully.
• Precautions are classified as DANGER, WARNING, or ATTENTION in order to clearly specify the level ofdanger and possible injury or damage that may occur if the device is misused.

Prohibition Do not use in the following cases or on the following areas.
Failure to comply may cause accidents or other problems.
• If your skin is damaged, experiencing a rash, inflammation or other abnormalities• If you are under the influence of alcohol• Head, anywhere inside the mouth, around the pubic region or any mucous membrane• Eyeballs • Upper eyelids (except in COOL Mode)• Lower eyelids, outer corner of the eye (except in EYE Mode and COOL Mode) • Throat• Bulging moles • Viral warts • Areas where you have broken a bone
• Areas of internal bleeding, swelling or scabbing after cosmetic medical procedures* ’Property damage’ refers to damage affecting dwellings, household belongings ordomestic animals and pets.

 Symbols
This symbol shows you what you must not do.

* Avoid using the device for 2 weeks after cosmetic medical procedures. After this period, consult with a physician
and only use the device if the physician confirms there is no problem.

(e.g. Hyaluronic acid injections, silicone injections, Botox, thread lifts, double eyelid surgery, HIFU, etc.)• Areas where metal was implanted by surgery (reconstructive surgery, plastic surgery, cosmetic
surgery)This symbol shows you what you must do.

This symbol shows you what you must be careful with.

DANGER

Prohibition

Do not use the device for a longtime over areas where the skin isthin or areas above bones.
Failure to comply may cause skinproblems or physical ailments.

Do not overuse the device.
For usage time and frequency,
follow the instructions in the User’s
Manual.

Prohibition Failure to comply may cause physicalinjuries.

The following individuals should not use the device:
Use of the device could cause electronic medical devices to malfunction.
 Do not use the device in conjunction with any medical electronic equipment, including:• pacemakers or other implanted electronic medical devices,
Prohibition• artificial heart-lung machines or other electronic life-support medical devices• electrocardiographs or other worn electronic medical devices.
1

Prohibition

Do not use continuously on thesame area except in COOL Mode.
Be sure to move the device acrossthe skin while it is in use. Do not use
on areas that cannot be reached by
the entire electrode head surface.
Failure to comply may cause burns or
low temperature injuries.

Prohibition

Before use, check to make sure theelectrode and treatment head are
not cracked, deformed or broken.
Do not use the device if they aredamaged or malfunctioning.
Failure to comply may cause injuries orskin problems.
2

WARNING

Do not use the device if the AC
adapter is damaged.
Contact your local distributor andProhibition use the dedicated power cord.
Do not leave the device
unattended while it is turned on.
Failure to comply may cause a fire
Prohibition or other accidents.

Non-intended
uses areprohibited

Do not use the device for any
purpose or areas other thanthose stated in the User’s
Manual.
Failure to comply may cause afailure of the device or an injury.

If any problem occurs during use,
including the presence of anabnormal odor or excess heat,
Prohibition against stop using the device.
use when an Failure to comply may cause theabnormal odor
or excess heat device to become damaged oris detected catch a fire.

3

Do not pull out a plug from a
power outlet or pull out the plugfrom the device during use.
Failure to comply may cause an
Prohibition electric shock or a fire due to a shortcircuit. Be sure to turn off the power
before pulling out the plug.

If the device is dropped ordamaged, turn off the power
immediately, pull out the plug,and request for a
repair/inspection.
Must Comply Failure to comply may lead to a fireor an electric shock.
Do not disassemble, attempt to
repair, or modify this device inany way.
Failure to comply may cause a fire,
Disassembly an electric shock, or other injuries.
is prohibited Contact your local distributor foradvice on repairing the device.
If you feel an earthquake, seelightning, or hear thunder, turn off
the power and unplug the device.
Removing Failure to comply may lead to a firethe plug or an electric shock.

Keep the device and AC adapter
away from children and do notallow them to use it alone. Do notallow children to play with thedevice.
User
restrictions Failure to comply may cause
accidents or injuries.

In the unlikely event that anabnormality occurs, immediately
unplug the AC adapter.
Do not use the device if smoke,
Disconnecting abnormal noises or an odor is
the AC adapter detected, or if it is otherwise
operating in an abnormal state.

Children should obtain theconsent of their guardians before
using the device and then use it
under the supervision of aguardian or a specialist.
User
restrictions Failure to comply may cause
accidents or injuries.

Do not damage, modify,
bundle, twist or otherwise apply
undue force to the AC adapter.
Failure to comply may cause theCareless or cable to break, resulting in a fire or
reckless use an electric shock. Straighten out
is prohibited
any twist in the cord before storing.

Do not use the AC adapter in thebathroom or areas where there iswater. Do not wash or immerse
the main unit or the AC adapter
Exposure in water.
to water Failure to comply may cause an
is prohibited electric shock, a short circuit or afire.

Do not use if the AC adapter isdamaged or the wall socket is tooloose.
Failure to comply may cause an
Use of a electric shock, short circuit, fire ordamaged failure of the device. Contact yourcable or plug local distributor for advice on
is prohibited
repairing the device.

When disconnecting the plugfrom a power outlet, pull it out by
grasping its head instead of thecord (do not touch the metalparts). Do not insert or
Removing disconnect the plug with wet or
the plug damp hands.
Failure to comply may cause anelectric shock, short circuit, fire or
failure of the device.

Safety Precautions

Safety Precautions

Prohibition

Do not use, store or charge the device in the following locations:• In bathrooms or near bathtubs, showers, washbasins or other containers filled with water• A dusty location, a location where there is exposure to direct sunlight or outdoors• A slope or a location where there are vibrations or shocks• A location where there are strong radio waves or magnetism• Near a flammable substance (such as alcohol, thinner, benzene, nail polish remover orspray).
• A location where there is exposure to oil smoke or steam• A location that may become hot, such as near a heating device (e.g. a heater)• A location with drastic temperature changes (e.g. near an air conditioner or heater)
Failure to comply may cause a fire, electric shocks, accidents, malfunctioning, falls or
injuries.

WARNING

After using the device and whencleaning it, be sure to turn off thepower and remove the plug fromthe wall socket.
Removing Failure to comply may cause thethe plug device to catch a fire or may lead to
accidents.
Always use the dedicated AC
adapter.
Failure to comply may cause an
Prohibition electric shock, short circuit or fire.
Do not exceed the rated capacityof a power outlet or wiring
accessory (do not connecting too
Overloading of many plugs in one outlet).
an electric Failure to comply may cause anelectric shock, short circuit, fire or
circuit isprohibited failure of the device.
Be sure to use in an operating
voltage range between
100~240 volts. Make sure thatthe plug is fully inserted into thepower outlet.
Must Comply
Failure to comply may cause a fire,electric shock or other accidents.

4Use in Foreign Countries
ATTENTION

Do not use while driving.
Failure to comply may cause
Prohibition accidents or injuries.
Do not use COOL Modecontinuously.
If the mode is used for continuous
treatments, an electrode head of thedevice will not become fully cool.
Prohibition Have an interval of at least 30
minutes between treatments.
Do not look directly into the LED
light. When using the LED light totreat upper eyelids, lower eyelids,
and outer corner of the eye, besure to close your eyes and keep
Prohibition an electrode head at least 5 cm
away from the skin.
Failure to comply may cause
damages to your eyes or skin.
Do not use at a temperature below
the specified ambient temperature.
Failure to comply may cause theProhibition electrodes to stick to the skin anddamage the skin.
If you are currently undergoing
dental treatment or have teeth
fillings, choose a lower intensity
level and apply the device on adifferent area,
Must Comply You may feel strong irritation or
discomfort by EMS on your cheeks
or chin. If you feel strong discomfort,
consult a physician.

5

If a power failure occurs, turn off
the power immediately and unplug
the device from the power outlet.
Removing Failure to comply may lead tothe plug accidents.

Do not allow the device to comeinto contact with another beauty
device or electronic appliance while
the power is on. Do not use withother electronic products (electric
Prohibition blankets/bed sheets, heated tables,etc.) at the same time.
Failure to comply may lead to
accidents due to malfunctions.
Use only Level 1 upon the first use
and until you are accustomed touse. Even if you do not feel any
stimulation, make only small
adjustments to the intensity level.
Use the device at an intensity level
Must Comply that suits you.
Make only gradual adjustments tothe intensity level. Failure to comply
may cause accidents or injuries.

WARNING
Use at AC100~240V only to avoid a potential fire or electrical shock hazard.
Must Comply

Failure to comply may cause a fire, electric shock or failure of the device.

Be sure to check the plug configuration and the voltage of the power outlet before use.

Use in Foreign Countries

Safety Precautions

Do not press the device strongly or
use it intensively on a single area.
Pressing the device stronger thannecessary or using it intensively on a
Prohibition
single area may damage the skin.

Combustion Failure to comply may cause a generation of heat, fire or malfunction of the device.
warning The plug configuration and voltage may vary depending on the region and the type of accommodation.
Always be sure to check for compatibility before using the device.
 For information about plug adapters, consult a travel agency or a consumer electronics retailer.

Check carefully that your skin is
not showing any abnormality
during the treatment. Stop using
the device immediately if you start
to feel pain, fatigue or if you see
any skin reaction during use.
Do not use the device
continuously for an overly longMust Comply period, such as for a commercial
use.
Stop using the device immediately if
the treatment does not agree withyour skin. If symptoms persist thefollowing day and beyond, consult aphysician.
Avoid using the device on the same
area for longer than 14 minutes.
When using on the same area
again, be sure to leave an interval of
24 or more hours between
treatments. Excessive use may
Must Comply damage the skin. The above
recommendations are guidelines
only; please observe your skincondition and use the device
accordingly.
6Names of Components

Before Use
How to operate/How to read the display

EYE Mode Electrodes Main Unit
<Front>
LED Lamps

After switching
to EYE Mode

<Rear>

Level Indicator(5 levels)

Level Indicator• Briefly press the button for the mode you wish tostart.

CLEAN Button / Level Button• For CLEAN and NEO DYHP Modes, there are 5
levels from which you can select.

NEO DYHP/EYE Button / Level Button• For EYE Mode, there are 3 levels from which you canselect.

COOL/POWER Button
Vibration (VIB) Button• There is no treatment level selection available for
COOL Mode.

Adapter Plug Insertion Port
Vibration (VIB) button[Accessories] Cotton Stopper• Connect the AC adapter to the adapter pluginsertion port on the main unit and the wall socket,
press and hold the POWER button to turn the deviceon (the device will be in Standby Mode).

Before Use

Features/Names of Components

Electrode Head

 Indicator and Buttons• For vibrations, you can select on/off and vibration
patterns using the Vibration (VIB) button.
There are 3 vibration patterns which you can select.
Please select the vibration pattern you prefer.
VIB 1 ⇒ VIB 2 ⇒ VIB 3 ⇒ OFF ⇒ VIB 1...

(Silver Ring)* The cotton stopper is attached to the electrode head
when purchased.
* Use this stopper to fix cotton pad to the electrode head.

 AC Adapter* The design and specifications of the device and accessories are subject to change for improvements without
prior notice.

78Modes
LED Lamps

Mode• Use only Level 1 upon the first use and until you are accustomed to use.

Button Display

Level Indicator
Button Display

StandbyLevel Indicator

CLEAN
CLEAN button lights up• The device and electrode head may become hot after using the COOL Mode.
Do not use COOL Mode continuously. Allow the device to rest (power-off status) for at least half an hour
after use.

LED Lamps

Features

Functions

Recommended Skin Lotion

Operating Time

Level

Frequency

Red LED turns on

Tones your skin and absorbs
dirt that cannot be removed
through normal facewashing.

• RF• EMS (high-frequency
waves)• Ion Cleansing• LED (Red)• Vibrations (ON/OFF)

RF Booster Pad*2
Pore Refining Deep Cleansing(or cotton pad with your regularskin lotion*1)

Approx. 3 minutes

5 levels

2 to 3 times a week

Orange and
Red LED turns on

Helps beauty ingredients topenetrate to the stratum
corneum and moisturizes
your skin.

• RF• NEO DYHP
• LED (Orange/Red)• Vibrations (ON/OFF)

RF Booster Pad*2
Clear Deep Moisturizer(If moisture is insufficient, add
the skin lotion remaining in thepouch.)

Approx. 5 minutes

5 levels

2 to 3 times a week

N/A

Improves the elasticity ofthe skin around the eyes byapplying weak
micro-currents.

• RF• Micro-currents• Vibrations (ON/OFF)Your regular eye-treatment serum*1

Approx. 3 minutes

3 levels

5 times a weekBlue LED turns on

Cooling treatment andlight (blue) improve skinelasticity and skin texture.

• COOL• LED (Blue)• Vibrations (ON/OFF)

Apply directly on the skin.

Approx. 3 minutes

N/A

Everyday use
is possible

1 - 5 lamps

NEO DYHPNEO DYHP button lights up• Use of a cotton pad in NEO DYHP Mode is optional.

*1. If using your regular skin careproducts, be sure that they arewater soluble. Do not use oil-basedor scrubbing skin care products.
Failure to comply may cause
malfunctions or skin problems.

1 - 5 lamps

Before Use

Before Use

3 buttons flashing• Be sure to move the device on your skin while using(except in COOL Mode).

EYE Standby

When rotating the electrode head 180°

NEO DYHP button flashing

EYE

When rotating the electrode head 180°

NEO DYHP button flashing rapidly

1 - 3 lamps

COOL* Only the central electrodehead has cooling functions

N/A

COOL button lights up

910

 Level Indicator

Pre-testing on Skin

There are 5 levels for CLEAN and NEO DYHP Modes, and 3 levels for EYE Mode.
Each time the CLEAN or NEO DYHP button is pressed, the level switches as shown in the below picture.
* There is no level selection available for COOL Mode.
* Use only Level 1 upon the first use and until you are accustomed to use.

* When using the device or a cosmetic product for the first time, always test for compatibility.

1

Clean your skin.
• Wash the inside of your wrist with soap, then wiping it dry with a towel.

CLEAN/NEO DYHP ModeB efo re UseLevel 1

Level 2

Level 3

Level 4

Level 5

• Do not wet or rub the pre-testing area.

B efo re Useskin-care product (gel or lotion) or water on your arm.
2 Apply
Use the device for approx. 5 minutes in NEO DYHP Mode.

EYE Mode

If you experience any skin abnormality (redness or burning sensation) at the pre-testingarea within 24 hours, stop using the device.

 How to Use EYE Mode
• This mode cares for the skin around the eyes using two electrodes on the electrode head.
• Rotate the electrode head 180° to switch between EYE Mode and Standby Mode.
The electrode head only rotates in the direction of the arrow shown in the figureon the right.
• The device cannot be operated in Eye Mode if the electrode head is not rotated 180°.
In addition, the device cannot be used in other modes while the electrode headis rotated.

EYE Mode Electrodes* *Rotates only as far
as 180°.

 Timer
One short beep will sound to indicate the half-way point in the operation time.
Refer to pages 9-10 for operation times in each mode.

 Standby Mode
When treatment on a certain mode is completed, two short beeps sound and the device switches to Standby
Mode (the state in which functioning is suspended). Refer to pages 9-10 for operation times in each mode.
Briefly press the Mode button again to start the selected mode.

11

ATTENTION• Be sure to perform a pre-test in order to prevent skin problems from occurring.
• If a rash, reddening, itchiness, stinging sensation or other abnormalities occur at the
pre-testing area, immediately stop the pre-test and wash the area.
• Check carefully to make sure there is no skin abnormality before using the device. Stop
using the device immediately if the treatment does not agree with your skin.

12How to Use
How to attach cotton pad to the electrode head1

Remove the cotton stopper from theelectrode head.

* If the cotton stopper does not fit properly on the electrode head, please use a larger and thinner
piece of cotton pad. Because thick cotton pads are difficult to fix in place, split them in half beforeuse.
* If the cotton pad is too thick, the thermal action of RF treatment may be adversely affected andyou will feel less heat.

3 Apply your regular skin lotion or gel onto the cotton pad.

* Make sure the skin lotion or gel coats the entire surface of the electrode head.
* If using a waterly (low-viscosity) skin lotion, the feeling of warmth may decrease.
Recommended skin lotions: RF Booster Pad or high-viscosity skin lotions.
* If the cotton pad becomes dry, add more skin-care product. Cotton pads may easily come offfrom the device when they are dry.

How to Use (General Instructions) Precautions
 Be sure to move the device on your skin while using. Using the device continuously on the same area
may cause burns or other injuries (except in COOL Mode).
 The warmth of the RF changes depending on the type of skin-care product used and the skin’s
condition at the time of treatment.
 Do not use this device around the eyes unless it is in EYE Mode or COOL Mode.
* The LED light can be used around the eyes and eyelids. Close your eyes and hold the device at least
5 cm away from the skin.
 Do not look directly into the LED light.
 The recommended ambient temperature is 15° C to 30°C.
 Make sure to remove makeup before using the device on your face. If you use the device over makeup,
the foundation may cause dark staining of the skin.

13 Treatment Method

1 Remove makeup and gently cleanse your face.

• Remove oil and dirt from your skin so that it is in a clean condition prior to treatment.
* COOL Mode can be used over makeup.

How to Use

How to Use

a piece of cotton pad on top of the electrode head
2 Placeand press the cotton stopper down over the cotton pad.

 Remove all metal accessories (watches, necklaces, rings, earrings, etc.) when using the device. Failure
to comply may cause accidents or injuries.
 Please reapply skin-care products or water during treatment to ensure that there is always some liquid
between the electrode head and the skin. If there is an insufficient amount of skin-care products orwater, the stimulation will weaken, causing skin problems.
 Please carefully clean the parts of the device that come into contact with the skin every time you use it
and keep the device clean at all times. Failure to comply may cause skin problems.
 If you feel an unpleasant tingling on your skin, reduce the intensity level or move to another area.
You could also moisten the skin with water, lotion, or gel before using the device.
Failure to comply may cause skin damage.

2 Attach a cotton pad if necessary and apply skin lotion onto it.
the AC adapter, press and hold the POWER button. Press the
3 Connectmode button and select a mode and intensity level.
• First, briefly press the button of the mode you wish to use to start. Then press the same button again
briefly to select the level.
• Start using the device in EYE Mode only after rotating the electrode head.
* There is no treatment level selection available for COOL Mode.
* Use only Level 1 upon the first use and until you are accustomed to use.

apply the treatment head to your skin and move the head over the
4 Gentlyarea being treated. Refer to pages 15 and 16 for more details.
• When operation in each mode is completed, the device will switch to Standby Mode. To start the next
mode, press the relevant mode button briefly.
* During use, make sure the electrode head is closely touching the skin.
* If you wish to stop the mode you are using half-way through the operation, press and hold the POWER
button.

and hold the POWER button to turn the power off. Disconnect the
5 Press
AC adapter from the main unit and the wall outlet.

6 After treatment, condition your skin as necessary.
• Please apply skin cream as appropriate.
* Clean the main unit after each use (page 17).

14

How to use the device on your skin
• Recommended speed for moving the device: approx. 2 cm per second* If the temperature feels too hot, move the electrode head more quickly.
• One short beep will sound to indicate the half-way point in the operation time.
When you hear the beeps , begin care on the other side of your face.
• Use the device in a position in which the entire surface of the electrode head comes in contact withyour skin.
• Be sure to move the device over the skin while using (except in COOL Mode).
Placing the device continuously on the same area may cause burns or low-temperature injuries.
• Do not use on the upper eyelids (except in COOL Mode), lower eyelids, or outer corner of the eye(except in EYE Mode and COOL Mode).

<NEO DYHP Mode>
Move the electrode head over the skinin the direction of the arrows shown below.

Press and move the device in a circular,
massaging motion as indicated by the arrows.
Areas you should not treat<CLEAN Mode>

How to Use

How to Use

Areas that can only be treated with LED light* Close your eyes and hold the device
at least 5 cm away from the skin.

Move the electrode head over the skin in the direction of the arrows shown below.
The LED light can be used to treat the upper eyelids, lower eyelids, and outer corner of the eye.

<EYE Mode>
Rotate the electrode head 180° so that the EYE Mode electrodes are at the top.
Make sure that both electrodes are touching the skin, press the electrode heads down repeatedly fromthe inner to the outer corner of the eye as if stamping the skin
EYE Mode Electrodes(Press → Release → Move electrode heads).
Press the electrodes down on the skin for approx. 3 seconds.
* Do not use the device on the eyelids.
* Be sure not to press down continuously on the skin for longer than 3 seconds.
Look in a mirror
while you are using
the device

Areas you should not treat<COOL Mode>
Areas you should not treat

Areas that can only be treated with LED light* Close your eyes and hold the device at least 5 cm away from the skin.

15

Press the electrode head repeatedly on your face as if stamping on the skin
(Press → Release → Move electrode head).
* The device can also be used over the lower eyelids and outer corner of the eye. Turn off the vibration function.
* When using the LED light on the upper eyelids, lower eyelids or outer corner of the eye, be sure to close your eyes
and keep the electrode head at least 5 cm away from the skin.
16Troubleshooting

After Treatment
How To Clean
To keep the product clean, please be sure to clean it after each use.

1
clean the device and electrode head, wipe them with a damp and2 Towell-wrung
cloth to remove any dirt. Wash the cotton stopper with water

Checklist for Possible Problems
Problems

Possible Causes

Solutions

Disconnect the AC adapter from the main unit and the wall outlet.

The AC adapter is not If you will not be using it for an extended period, clean the device before storing it.
 When the device is not in use, disconnect the AC adapter from the main unit and wall outlet.
 Avoid storing the device near a heater or in a bathroom or other high-temperature/high-humidity
locations. Also, avoid locations where there is an exposure to direct sunlight or unstable.
 Keep the device out of the reach of children.

The device is too warm.

17→

Plug adapter into the device.

→

The device was used
continuously.

→

Turn off the device and wait
at least for 1 hour before
restarting to use.

→

EYE Mode will not function
unless the EYE Modeelectrodes are properly rotated
180°. Rotate the electrodes sothat they are in the correct
position (page 11).

→The EYE Mode electrodes
are not touching the skin.

→

Please make sure both of thetwo EYE Mode electrodes
are touching the skin at thesame time (see page 16).

→Vibrations are turned off.

→

Press the Vibration (VIB)
button to switch on thevibration function.

→

COOL Mode was used
continuously.

→

Turn off the device and wait
at least for 30 minutes
before restarting to use.

→

The device is being used in
an environment with atemperature exceeding therecommended range.

→

Try using in an environment
with the temperature
between 15°C and 30°C.

→

The outside of the electrodehead will not cool down.

→

Only the inner electrodehead becomes cool.

The electrode head is not in

EYE Mode does not
function.

You cannot feel any
vibration.

 If you will not use the device for an extended period, be sure to inspect it carefully and check that the device
works properly and safely before restarting use.

 If the device malfunctions or appears damaged, please contact your local distributor.

The adapter is not plugged
into the device.

→ the correct position.

Maintenance and Inspections

 If the device is malfunctioning, go to ‘Checklist for Possible Problems’ (pages 18-19) and attempt to identify
the problem before arranging to have the device repaired. If the problem persists, contact your local
distributor with a proof of purchase.

→

COOL Mode is not very cool.

Troubleshooting

After Treatment• Refrain from using an alkali cleaning agent, alcohol, thinner, benzene, nail polish remover or any
other solvents.
• Do not leave the device with lotion or gel on it.

StorageTurn on the power by
connecting the AC adapter.

The device does not turn on.

after removing it from the main unit.

the entire device with a dry cloth and store it after removing all
3 Wipemoisture.

→ plugged into the wall socket. →18

Problems

EMS causes uncomfortabletingling or pain.

You cannot feel the EMS or
it feels too weak

You cannot feel RF warmth.

19→

The level setting is too high.

Solutions→

Lower the level using theLevel button.

→

The entire surface of theelectrode head is nottouching the skin.

→

Use a mirror to check thatthe entire surface of theelectrode head is touching
the skin.

→

The level setting is too low.

→

Increase the level with Level
button.

→The skin lacks moisture oris dry.

→Moisturize your skin withlotion or gel before use.

→

The electrode head is not
moved along the muscles.

→

Apply the electrode headalong muscles (page 15).

→

You are using the device
without moving the electrode
head.

→

Be sure to move the electrodehead over your skin whenusing the device, except in
COOL Mode.

→

You are moving the device
too slowly on the skin.

→Move the device a little
quicker.

→

It may feel very hot
depending on the skin-care
product used or thepresence/absence of a cotton
pad.

→

Use the device without acotton pad.

→Change the skin-care
product used.

→

Standby Mode has been
activated (page 11)→

Press the relevant mode
button to start the mode.

→

The skin lacks moisture or is
dry.

→

Moisturize your skin withlotion or gel before use.

→

You are moving the device
too quickly.

→

Please move the device alittle slower.

Frequently Asked Questions
Q

A

How much lotion or gel
should I use each time?

This device is designed to achieve better results with sufficient useof lotions and/or gel.
As a general guide, when using cotton pads, be sure that theentire electrode head is moistened with face lotion.
Use about two pearl-sized drops of gel.
We recommend adding more lotion or gel if you feel there is notenough.

Can I use it over a facialmask?

It can be used over a facial mask too (except in CLEAN Mode).

I have a tingly sensation
on my skin.

This tingly sensation is caused by electrical stimulation from
Ion/EMS/NEO DYHP.
It is possible to adjust the level using the Level button. Use thedevice at a strength that feels comfortable for you.
Use only Level 1 upon the first use and until you are accustomed
to use.

Why do I not feel warmth of
RF when using CLEAN,
NEO DYHP or EYE Modes?

Move the electrode head more slowly or press harder against theskin surface. It is also possible that the moisture absorbed in the
cotton is insufficient. Add more lotion or gel.

Why can’t I see any dirton the cotton pad?

Dirt is removed even though it may not be visible on the cotton.
The amount of visible dirt on the cotton can vary amongindividuals.

The electrode head does notwarm up when I use RF Mode.

The RF does not warm the electrode head. Rather, it uses thepower of electricity to warm your skin.

What is the recommended
frequency and duration of use?

Recommended frequency of use:
2 to 3 times a week for Clean/NEO DYHP Mode. 5 times a week
for EYE Mode. The device can be used daily for COOL Mode.
Recommended duration of use:
CLEAN, EYE, COOL Modes: approx. 3 minutes. NEO DYHP
Mode: approx. 5 minutes. Excessive use may damage the skin.
The above recommendations are guidelines only; please observe
your skin condition and use the device accordingly.

Troubleshooting

Troubleshooting

The treatment temperature
is too high (except in COOL
Mode).

Possible Causes

20After-sales Service
 Product Specifications
 Disposing of the device

Rated Power Supply

DC12V 5.0A

• Follow your local government rules for proper disposal of this product.

Power Consumption

Approx. 20 W

Outer Dimensions

Main unit (with a cotton stopper): Approx. W 50 x D 65 x H 185 (mm) Contacts for overseas customers

WeightMain unit (with a cotton stopper): Approx. 230 g

Materials

Main unit: ABS resin Panel: Polycarbonate
Electrode head: PET resin and stainless steel

Please contact your local distributor or visit YA-MAN Global websites (https://www.ya-man.com/en/).

Correct Disposal of this product● This marking indicates that this product should not be disposed with other household wastes throughout

the UK and EU. To prevent possible harm to the environment or human health from uncontrolled waste
disposal, recycle it responsibly to promote the sustainable reuse of material resources. To return your useddevice, please use the return and collection systems or contact the retailer where the product waspurchased. They can take this product for environmental safe recycling.

Country of manufacture

Japan

Accessories

Cotton stopper, AC adapter

Recommended
Environmental
Temperature

15°C to 30°C

After-sales Service
21

22