Features

ISOVOLT
Titan|neoRobust, reliable
stationary X-ray
generators
The ISOVOLT Titan|neo generator powers a range ofradiographic inspection technology—delivering the mostreliable, consistent results in even the highest accuracy
applications. So you can increase precision and inspectmultiple parts each day even in 24/7 Testing Machines
applications.
Increased reproducibility: Reduce exposure times forvarious materials in several operation modes with high,stable radiation and fluctuations <0.05%.
Greater dependability: An extended tube range andmaximum current ensure enhanced imaging contrast
and high penetration power.
Unmatched flexibility: Its modular design includes
intelligent tube integration and permanent systemmonitoring—offering unmatched ramp-up times* anda 100% duty cycle for continuous operation in inline
systems**.

Powerful performance

Permanent system monitoring

Modular design for easy integration

Convenient, user-friendly controls

Built-in safety features*Depending on permissible tube data.
**Subject to operational generator cooling.

Technical specifications
High voltage generator

HP160

HP225

Max out current mA45

45

Max out power kW

4.5 (limited by tube spec; 1 phase mains:4.0 kW)

4.5 (limited by tube spec; 1 phase mains:4.0 kW)

Insulation

Oil

Oil

Housing dim (w x d x h)

340 x 945 x 750 mm(13.38” x 37.20” x 29.52”)

340 x 945 x 750 mm(13.38” x 37.20” x 29.52”)

Weight195 kg (429.9 lbs)

190 kg (418.87 lbs)

Presel and settings

From 5 to 160 kV in 1 kV

From 5 to 225 kV in 1 kV

Dig display of set and act values

3 digits (set); 4 digits (act)

3 digits (set) ; 4 digits (act)

Display resolution

1 kV (set); 0.1 kV (act)

1 kV (set) ; 0.1 kV (act)

Accuracy<1%<1%

Reproducibility<0.01%<0.01%

Temperature drift<80 ppm/K<100 ppm/K

Presel and settings

From 0.1 to 45 mA in 0.1 mA

From 0.1 to 45 mA in 0.1 mA

Dig display of set and act values

3 digits

3 digits

Display resolution

0.1 mA

0.1 mA

Accuracy<1%<1%

Reproducibility<0.25%<0.25%

Temperature drift<100 ppm/K<100 ppm/K

Programmable timer

11Presel and setting

1 ... 9999 s

1 ... 9999 s

Dig display of set and act values

4 digits

4 digits

Prewarning

Audible and visible

Audible and visible

Presel and setting

2 ... 120 s or deactivated

2 ... 120 s or deactivatedNumber of storable programs

250250Warm-up

Auto mode based on real time clock

Auto mode based on real time clock

X-ray tube set up

Tube selectable from a database of morethan 20 pre-programmed, unipolar tubes

Tube selectable from a database of morethan 20 pre-programmed, unipolar tubes

Operation history

Stored on SD

Stored on SD

Warm-up history

Stored on SD

Stored on SD

Dimension wxdxh

440 x 114 x 295 mm (17.32” x 4.48” x 11.61”)

440 x 114 x 295 mm (17.32” x 4.48” x 11.61”)

Weight3.8 kg (8.37 lbs)

3.8 kg (8.37 lbs)

AUX: 1N PE 230 V ±10% 50/60 Hz 10 A,

AUX: 1N PE 230 V ±10% 50/60 Hz 10 A,

MAIN: 3N PE 400/230 V ±10% 50/60 Hz 20 A
or 1N PE 230 V ±10% 50/60 Hz 63 A,

MAIN: 3N PE 400/230 V ±10% 50/60 Hz 20 A
or 1N PE 230 V ±10% 50/60 Hz 63 A,

3-phase, grounded neutral TN-S or TN-C-S
mains (star connected system), optional
3-phase isolation transformer
Separate grounding for X-ray tube and highvoltage generator (minimum 6 mm²)
AUX: 10 A (1N PE)

3-phase, grounded neutral TN-S or TN-C-S
mains (star connected system), optional
3-phase isolation transformer
Separate grounding for X-ray tube and highvoltage generator (minimum 6 mm²)
AUX: 10 A (1N PE)

MAIN: 63 A (1N PE) or 20 A (3N PE)
Time-delay fuses, customer-supplied

MAIN: 63 A (1N PE) or 20 A (3N PE)
Time-delay fuses, customer-supplied

Operating temperature range

0 °C to +40 °C

0 °C to +40 °C

Storage temperature range-30 °C to +70 °C-30 °C to +70 °C

Max out voltage kV

Tube voltage

Tube current

Exposure time

Programmed mode

Control module

Connected loads

Power connection

Grounding
Mains fuses

160225

High voltage generator

HP320

HP450

HR240

Max out voltage kV

320450240

Max out current mA

4545

3

Max out power kW

4.5 (limited by tube spec;
1 phase mains: 3.5 kW)

4.5 (limited by tube spec;
1 phase mains: 3.5 kW)

0.320 (limited by tube spec)

Insulation

Oil

Oil

Oil

Housing dim (w x d x h)

340 x 945 x 750 + 340 x 945
x 540 mm(13.38” x 37.20” x 29.52”) +13.38” x 37.20” x 21.25”)

340 x 945 x 750 + 340 x 945
x 540 mm(13.38” x 37.20” x 29.52”) +13.38” x 37.20” x 21.25”)

340 x 945 x 750 mm(13.38” x 37.20” x 29.52”)

Weight190+140 kg(418.87 + 308.64 lbs)

190+140 kg(418.87 + 308.64 lbs)

170 kg (374.78 lbs)

Presel and settings

From 10 to 320 kV in 1 kV

From 10 to 450 kV in 1 kV

From 5 to 240 kV in 1 kV

Dig display of set and actvalues

3 digits (set); 4 digits (act)

3 digits (set); 4 digits (act)

3 digits

Display resolution

1 kV (set); 0.1 kV (act)

1 kV (set); 0.1 kV (act)

1 kV

Accuracy

<1%

<1%

<1%

Reproducibility<0.01%<0.01%<0.01%

Temperature drift<80 ppm/K<80 ppm/K<80 ppm/K

Presel and settings

From 0.1 to 45 mA in 0.1 mA

From 0.1 to 45 mA in 0.1 mA

From 0.01 to 3 mA in
0.001 mA

Dig display of set and actvalues

3 digits

3 digits

4 digits

Display resolution

0.1 mA

0.1 mA

0.001 mA

Accuracy<1%<1%<1%Reproducibility<0.25%<0.25%<0.25%

Temperature drift<100 ppm/K<100 ppm/K<100 ppm/K

Programmable timer

11

1

Presel and setting

1 ... 9999 s

1 ... 9999 s

1 ... 32767 s (xs-control)

Dig display of set and act
values4 digits

4 digits

5 digits

Prewarning

Audible and visible

Audible and visible

Audible and visible

Presel and setting

2 ... 120 s or deactivated

2 ... 120 s or deactivated

2 ... 255 s or deactivatedNumber of storable
programs

250250—Warm-up

Auto mode basedon real time clock

Auto mode basedon real time clock

Automated intelligent
tube conditioning

X-ray tube set up

Tube selectable from a
database of more than
25 pre-programmed,
bipolar tubes

Tube selectable from a
database of more than
25 pre-programmed,
bipolar tubes—

Operation history

Stored on SD

Stored on SD—

Warm-up history

Stored on SD

Stored on SD—

Tube voltage

Tube current

Exposure time

Programmed mode

Control module
Dimension (w x d x h)

440 x 114 x 295 mm(17.32” x 4.48” x 11.61”)

440 x 114 x 295 mm(17.32” x 4.48” x 11.61”)—Weight

3.8 kg (8.37 lbs)

3.8 kg (8.37 lbs)—

AUX: 1N PE 230 V ±10% 50/60
Hz 10 A,

AUX: 1N PE 230 V ±10% 50/60
Hz 10 A,

MAIN: 3N PE 400/230 V ±10%50/60 Hz 20 A or 1N PE 230 V±10% 50/60 Hz 63 A,

MAIN: 3N PE 400/230 V ±10%50/60 Hz 20 A or 1N PE 230 V±10% 50/60 Hz 63 A,

3-phase, grounded neutral
TN-S or TN-C-S mains(star connected system),
optional 3-phase isolationtransformer

3-phase, grounded neutral
TN-S or TN-C-S mains(star connected system),
optional 3-phase isolationtransformer

Connected loads

Power connection

Grounding

1N PE 230 V ± 10% 50/60 HZ
10 A AUX,
1N PE 230 V ± 10% 50/60 HZ
10 A MAIN

Separate grounding for
Separate grounding for
Separate grounding for
X-ray tube and high voltage X-ray tube and high voltage X-ray tube and high voltage
generator (min. 6 mm2)
generator (min. 6 mm2)
generator (min. 6 mm2)
AUX: 10 A (1N PE)

AUX: 10 A (1N PE)

MAIN: 63 A (1N PE) or 20 A
(3N PE) time-delay fuses,
customer-supplied

MAIN: 63 A (1N PE) or 20 A(3N PE) time-delay fuses,
customer-supplied

Operating temperature
range

0 °C to +40 °C

0 °C to +40 °C

0 °C to +40 °C

Storage temperature range-30 °C to +70 °C-30 °C to +70 °C-30 °C to +70 °C

Mains fuses

10 A (1N PE) integrated
into aux switch,
10 A (1N PE) integrated
into main switch

Waygate Technologies
Bogenstr. 4122926 Ahrensburg
Germany
Tel.: +49 4102 807 0Fax: +49 4102 807 189
E-mail: xray.info@bakerhughes.com
Waygate Technologies
201 Beltway Green Blvd.
Pasadena, Texas 77503
Tel.: +1 281 542 3600

Copyright 2021 Baker Hughes Company. All rights reserved.

BHFF34152EN (01/2023)

waygate-tech.com