Precision Nitrogen Trace
Nitrogen Generator for GC
Part Number : See Reverse

Service Kit : 08-3613

Your local gas generation partner

Description
Precision Nitrogen Trace has been developed to provide a constantand consistent source of zero nitrogen for carrier and detector gas
for GC applications. These generators are capable of delivering
ultra high purity nitrogen, removing oxygen and moisture via
Pressure Swing Adsorption technology, as well as removinghydrocarbons by means of catalytic oxidation to ensure maximum
purity output.
As with all Precision Series generators, Nitrogen Trace models
benefit from a compact and modular, stackable design,minimizing the total footprint required for GC gas supply, andproviding flexibility to add or remove modules as your laboratory
requirements evolve over time.

Applications)

Sample Prep
• Headspace• Purge and Trap
• Tube Conditioning)

Carrier Gas•GC)

Detector Gas• Make-up / Reference

406 mm

540 mmKey Features)

Suitable for carrier gas and detector gas)

Generates zero nitrogen on demand fromcompressed air)Regenerative CMS columns remove oxygen andmoisture)

Catalyst chamber to remove hydrocarbons (asmethane) to market- leading purity of <0.05ppm)

Ultra-fast start-up time, quick to reach standard
operating purity)

Minimum maintenance with an annual filter change)

Compact, space-saving modular design

380 mmPrecision Nitrogen Traceshown in a typical Precision stack.
Copyright © Peak Scientific Instruments Ltd 2024 — Precision Nitrogen Trace Data Sheet Rev 10 RSID 385 EN —08/07/2024

Technical Specifications

Nitrogen Trace, 250cc

Nitrogen Trace, 600cc

Nitrogen Trace, 1000cc

250cc/min

600cc/min

1000cc/min

Max Flow Rate
Max Pressure80 psi / 5.5 bar

Purity>99.9995%Hydrocarbon Content<0.05 ppm

Gas Outlets

1 x 1/4" BSPP

Min/Max Air Inlet Pressure120-145 psi / 8.3-10 bar

Min Air Inlet Flow18 lpm

Min Inlet Air Quality

22 lpmISO8573 - 1:2010 Class 1.4.1

Phthalates

None

Suspended Liquids

None

Start-Up Time For Purity

1.5 Hours

Operating Temperature

5°C - 35°C / 41°F - 95°F

Electrical Requirements

110VAC ± 10%/60 Hz/4.4A or 230VAC ± 10%/50 Hz/2.1A

Power Consumption

484 Watts

Heat Output

up to 1650 BTU/HrGenerator Dimensions (WxDxH)

380 x 540 x 406mm / 14.9 x 21.2 x 15.9"

Generator Weight

38 Kg / 83.6 lbs

Noise Level

Silent in operation

Ordering Information

110V

230V

110V

230V

110V

230V

Part Number

62-0251

62-0250

62-0601

62-0600

62-1001

62-1000

Annual Service
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan

Precision Air Compressor

110v

230v

Annual Service Kit

Part Number

65-1555

65-0555

08-8372

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management Systemconforms to: ISO:9001:2008

For a full list of our worldwideoffice locations, please visit:
Web: www.peakscientific.com
Email: discover@peakscientific.com
Copyright © Peak Scientific Instruments Ltd 2024 — Precision Nitrogen Trace Data Sheet Rev 10 RSID 385 EN —08/07/2024