Product Datasheet

Arium®
Smart Station
Flexible Remote Dispenser---Advantages
Compact: Save space integrating in your lab
Intuitive: touch-activated color display with direct accessto all important functions
Flexible: Stepless height adjustment to fill different
size containers
Ergonomic Design: Hand-held for optimized leftor right-hand operation
Accurate: Precise volume dispense for reliable bufferand sample preparation

Product Description
The Arium® Smart Station is designed for flexible remote
dispensing of pure and ultrapure water directly at the pointof use.
While dispensing water into a broad range of different sized
containers, the Smart Station offers constant control of allimportant quality parameters, at all times.
The ergonomic design supports left and right hand operationand can be easily adjusted to your need.

CompactIntuitive

The Arium® Smart Station will readily fit under almost every
standard laboratory shelf. This handy device will give theflexibility you need to integrate it into any lab environmentand save space.

Easy to operate the Arium® Smart Station using thetouch-activated color display, even when you are wearing
laboratory gloves. The icons will guide you through themenu for intuitive, error-free operation.

Accurate

Flexible

Optimize your sample preparation and benefit from directaccess to all important dispensing functions: manual,volume- controlled or predefined favorite volumes.
The various dispensing functions offers precise dispensing
from 2 L/min up to manually controlled drop-by-drop*
function.

The stepless height adjustment from 8.5 – 65 cm allows
you to fill different sized containers. Thanks to its angulardesigned stand even bulky vessels can be optimally
positioned under the dispense outlet. Depending on
your needs you can adjust the volume dispense from
50 mL up to 50 Liter.

Choose the Smart Station of Your Choice
The Smart Station is designed to be used for ultrapure and pure water dispense connected to yourArium® Ultrapure water system or Bagtank:Type of System

Use withArium® Smart Station Ultrapure

Direct connection to Arium® Pro or Arium® Comfort supplying ultrapure Type 1 waterArium® Smart Station Pure

Direct connection to Bagtank supplying pure water*Depends on the connected system type

Supply ultrapure water from Arium® Pro and Arium® Comfort2

Supply pure water from Arium® Bagtank

Adapt the Dispense Point to Your Needs
To optimally fit the Smart Station into your lab environment
there are two installation options available: bench top orwall-mounted.
Both options can be individually adjusted to left or righthand side operation, without any additional required
accessories.

Flexible Dispensing
Connect up to three Smart Stations to your single ultrapure water system to serve different dispensing areas. By using
the extended connection set, the distance between each Smart Stations can be set up to 4 meters. Adapted to yourapplication you could select between different point of use filters.

3Technical Specifications
Smart Station Bench-Top
Dimensions:Control box with stand (w × d × h)

213 × 213 × 598 mm (8.4 × 8.4 × 8.2”)

Operating range fixed dispense arm (d × w × h)

428 × 476 × 835 mm (16.9 × 18.7 × 32.9”)

Tubing Length:
Distance to water system/bagtank

2 Meter

Operating range flexible hand held

0.7 Meter

WeightApprox. 4.9 kg (10.8 Ibs)

Smart Station Wall-Mounted
Dimensions:Control box (w × d × h)

172 × 157 × 343 mm (6.8 × 6.2 × 13.5”)

Operating range dispense arm (d × w × h)

242 × 90 × 300 mm (9.5 × 3.5× 11.8”)

Tubing Length:
Distance to water system/bagtank

2 Meter

Operating range flexible hand held

0.7 Meter

WeightApprox. 2.4 kg (5.3 Ibs)

General Specifications
Volume-controlled dispensing

50 mL – 50 Liter

Volume accuracy± 5 % between

Power supply

100 – 240 VAC; 50 and 60 Hz, 2.5 A (max.) 2°C – 40°C

Power cord (IEC 60320-1 / C14)Country specificOrdering Information
Order numberDescription

H2O-ARST-UP-T

Arium® Smart Station Ultrapure for benchtop instatllationH2O-ARST-UP-B

Arium® Smart Station Ultrapure for wall-mounted instatllation

H2O-ARST-P-T

Arium® Smart Station Pure for benchtop instatllation

H2O-ARST-P-B

Arium® Smart Station Pure for wall-mounted instatllation

Benchtop and wall-mounted edition can be assembeled for left or right hand side, without additional equipment required.

4

Consumables
Arium® Sterile Plus-----

For Sterile and Particle-Free Water Dispensing
Excellent total throughput and flow rates
Integrity-tested
Validated according to HIMA and ASTM F-838-05
Meets WFI quality standards compliant with USP,
incl. USP Plastic Class VI Test
Manufactured according to EN ISO 9001
Easy to install
Certified quality

Technical Specifications
Materials
Membrane

Asymmetrical polyethersulfone

Filling bell assembly

Polycarbonate

Other plastics

Polypropylene

Typical Specifications for Purified Water
Flow rate(depends on the inlet pressure
and the type of system)

Up to 2.0 L/min.

Bacteria< 0.001 CFU/mL

Particle content

No particles > 0.22 μm

General Specifications
Pore sizes

0.45 μm + 0.2 μm

Filtration area

0.015 m²

Inlet and outlet¼” plug-in connector

Sterilization(3 cycles max.)

Autoclavable at 134°C,
1 bar (14.5 psi), 30 min.

Max. diffusion

1 mL/min. at 2.5 bar (36 psi)

Min. bubble point

3.2 bar (46.4 psi)

Description
The Arium® Sterile Plus (Sartopore® 2 150) is a sterile,
ready-to-use membrane filter capsule designed to meetthe highest requirements.
The filter capsules contain a hydrophilic, heterogeneouspolyethersulfone double-layer membane, delivering
excellent total throughput and flow rate performance.
The capsule is attached by a quick-connect coupling to
the point of use and reliably removes particles and micro­organisms > 0.2 μm in the last water purification step.
The pleated Sterile Plus membrane filter units are validated
as sterilizing-grade filters for biopharmaceutical use
according to HIMA and ASTM F-838-05 guidelines(documents available). Each capsule is integrity-tested
during the manufacturing process to ensure it meets thehighest quality standards and safety regulations.

Ordering Information
Order Number

Description

5441307H4--CE

Arium® Sterile Plus
(Sartopore® 2 150 capsule),
0.2 μm pore size; qty. per pkg.: 1--

Intended Use
Device Type:Arium® Smart Station Ultrapure
Arium® Smart Station Pure

5Arium® Cell Plus Ultrafilter----For Effective Removal of Endotoxins inCell Culture Applications
Effective removal of RNase | DNase
Reliable removal of endotoxins
High flow rate performance
Certified quality
Sterile-packaged

Technical Specifications
Materials
Membrane

Polysulfone

Composite material

Polyurethane (PUR)

Housing

Acrylonitrile butadiene styrene (ABS)Protective bell

Polycarbonate (PC)Typical Specifications for Purified Water
Flow rate(depends on the inlet pressure
and the type of system)

Up to 2.0 L/min.

Endotoxins< 0.001 EU/mLBacteria< 0.01 CFU/mL

RNase concentration< 1 pg/mL

DNase concentration< 5 pg/mL

General Specifications

Description
The Arium® Cell Plus is a point-of-use ultrafilter for theeffective removal of small molecules at 15kDa cut off.
Attached with the quick-connect coupling to the pointof use, the ultrafilter provides the highest safety for yourcritical cell culture and life science applications byremoving high efficiently RNase | DNase and Endotoxins.
A protective bell supplied with the ultrafilter additionallyprevents retrograde contamination.
Moreover, the high-quality material selected for Arium® CellPlus enables excellent total throughputs and optimal flow
rates.

6Cut-off

15.000 Daltons/0.005 μm

Inlet and outlet¼” plug-in connector

Dimensions (height × diameter)

169 mm × 50 mm (6.6” × 1.9”)

Max. operating pressure

6 bar (87 psi)

Max. inlet temperature

50°C

Effective membrane area

0.5 m² (5.4 ft²)

Ordering Information
Order Number

Description

H2O-CUF

Arium® Cell Plus;
qty. per pkg.: 1--

Intended Use
Device Type:Arium® Smart Station Ultrapure
Arium® Smart Station PureAccessories
Arium® ExtendedConnection Set--

Flexible enhance your working space
Extend the operating range up to 4 meter
Fast and reliable connnection

Technical Specifications
Materials
Cable

Ethernet CAT6A

Tubing

1/4“ Polyethylene (PVDF | PE)

LengthOperating range

4 meter

Ordering Information

Description
The extended tubing set is used for the Smart Station
Ultrapure and Pure for uncomplicated extension of thestandard supplied tubing set.

Order Number

Description

H2O-ATES-UP

Arium® Tubing Extension Set
Ultrapure

H2O-ATES-P

Arium® Tubing Extension Set Pure--Intended Use
Device Type:Arium® Smart Station Ultrapure
Arium® Smart Station Pure

With the extension set the operating range can be easily
increased from 2 up to 4 meter.

7

Sartorius Service
We Ensure the Quality of Your Results

Our Services at a Glance

At Sartorius, quality products go hand in hand withprofessional service. With our wide service offering, we will
help guarantee the safe, reliable and optimal operation ofyour Arium® system. Just ask us and we will even cover theentire life cycle of your laboratory water system – fromcommissioning to qualification to regular maintenance.
Together with you, we will ensure the consistently highquality of your laboratory water purification.

Installation and Commissioning
Your advantage: Your system will operate reliably at peak
performance from day one
Equipment Qualification (IQ | OQ)Your advantage: You will meet all regulatory requirements(GMP | GLP)
Regular Preventative Maintenance, Including Calibration,
inspection and testing of your system and exchange ofconsumables
Your advantages: Optimal operation of your system; reliableresults; prevention of downtime or even equipment failure
Get more information now at:www.sartorius.com/service

Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Straße 2037079 Göttingen
Phone +49 551 308 0F or further information, visit
www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 04 | 2021

USASartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906