R

User Guide
Elcometer 456
Scan Probes

www.elcometer.com

R

CONTENTS
SectionOverview
1
2
Calibration Using a Scan Probe
3
Technical Specification
4
Spares5
Important Information

Page
en-2
en-3
en-4
en-5
en-5

For the avoidance of doubt, please refer to the original English language version.
This product is packed in a cardboard package. Please ensure that all packaging is disposed of in anenvironmentally sensitive manner. Consult your local Environmental Authority for further guidance.
is a registered trademark of Elcometer Limited, Edge Lane, Manchester, M43 6BU. United Kingdom
All other trademarks acknowledged.
© Elcometer Limited 2008 - 2018. All rights reserved. No part of this document may be reproduced, transmitted,
transcribed, stored (in a retrieval system or otherwise) or translated into any language, in any form or by anymeans (electronic, mechanical, magnetic, optical, manual or otherwise) without the prior written permission of
Elcometer Limited.
R

en-1

www.elcometer.com

R

1 OVERVIEW
Elcometer 456 Scan Probes have been specifically designed for use
with the “Scan” and “Auto Repeat” modes on Elcometer 456 Model T
separate gauges. They are supplied with a highly durable ‘snap on’,user replaceable, probe capa - allowing users to take individual
readings or rapidly scan large surface areas without damaging theprobe or the coating.
Scan Mode: When Scan Mode is selected,
users can slide the Scan Probe over theentire surface area. As the probe is lifted
off the surface the gauge displays theaverage coating thickness value (x)together with the highest (Hi) and lowest(Lo) thickness values. Each set of three
readings can be displayed on a run graphand stored into the memory. This is ideal foron line inspection.
Auto Repeat Mode: When the probe isplaced onto the coated substrate in Auto
Repeat Mode, the gauge automaticallytakes readings at a rate of 140 per minuteuntil the probe is removed from thesubstrate. Each individual reading isstored into the memory. The Auto Repeat
Mode can be used to significantly speed
up the inspection of large coated areas
such as steel I-beams (RSJ’s), tanks, ship
hulls, etc.

Hi, x & Lo

1, 2, .......nNote: “Scan Mode with Hold” allows the probe to be lifted off the surface for a
maximum of 1.5 seconds and replaced without starting a new set of readings allowing the user to avoid holes, weld seams, etc. in the substrate without
affecting the measurement results.

aReplacement probe caps are available - see Section 4 “Spares” on page en-5 for details.

www.elcometer.com

en-2

R

2 CALIBRATION USING A SCAN PROBE
When using a Scan Probe in “Scan” or “Auto Repeat” modeb, theprobe cap must be fitted to avoid damage to the probe tip and thegauge can only be calibrated using either the “Smooth” or “Rough /2 Point” calibration methods.
Scale 1: For “Smooth” calibration a 1500µm (60mils) nominal foil anduncoated base / zero plate is required. For “Rough / 2 Point” a 50µm
(2mil) nominal foil is also required.
Scale 2: For “Smooth” calibration a 4mm (160mils) nominal foilc anduncoated base / zero plate is required. For “Rough / 2 Point” a 50µm
(2mils) nominal foil is also required.
2.1 CONNECTING THE SCAN PROBE
1 Rotate the probe plug to align thepins.
2 Screw in the collar - clockwise.
2.2 CALIBRATING WHEN IN “SCAN”
OR “AUTO REPEAT” MODEb:
1 Press the Cal softkey and select“Cal Method”.
2 Select “Smooth” or “Rough /2 Point” followed by “Calibrate” d.
3 Follow the on screen instructions.

2

1

In order to avoid probe tip damage during “Scan” or “Auto Repeat”
mode, a probe cap must be fitted. During the calibration procedure if,
when asked by the gauge, no cap is to be used, the gauge willautomatically revert to “Standard” measurement mode.
Scan Probes use the Elcometer 456's patented offset feature ensuringthat any cap wear during use is incorporated within the calibration
procedure, displaying the maximum coating thickness range of theprobe and cape combination at the end of the calibration routine.
b

To select the measurement mode go to Menu/Setup/Measurement Mode.
The Scale 2 probe is supplied with a 3mm (120mils) and 1mm (40mils) foil which can be used to make a
4mm (160mils) foil.
d
If an alternative calibration method has been selected, the gauge will prompt the user to select either“Smooth” or “Rough / 2 Point”.
eProbe caps have a nominal thickness of 500µm (20mils).
c

en-3

www.elcometer.com

R

3 TECHNICAL SPECIFICATION
3.1 PROBE CAP FITTED
Range

Resolution

Accuracy

Scale 1100 - 1000µm (4 - 40mils)

Scale 2

0.1 - 4.5mm (4 - 180mils)

Scale 11µm (0.1mil)

Scale 21μm: 0 - 1mm; 10μm: 1 - 5mm(0.1mil: 0 - 50mils; 1mil: 50 - 200mils)

Scale 1

±3% or ±2.5µm (±0.1mil)f

Scale 2

±1-3% or ±0.02mm (±1mil)f

Part Number

Probe Design

Minimum
Headroom

Minimum
Sample Diameter

T456CF1U

Ferrous Scale 1

86mm (3.38")15mm (0.59")

T456CF1UARM

Ferrous Scale 1 Armoured Cable

140mm (5.51")

15mm (0.59")

T456CFNF1U

Dual FNF Scale 1

89mm (3.50")

15mm (0.59")

T456CF2U

Ferrous Scale 2

90mm (3.54")15mm (0.59")

3.2 WITHOUT PROBE CAP
Range

Resolution

AccuracyfgScale 1

0 - 1500µm (0 - 60mils)

Scale 20 - 5mm (0 - 200mils)

Scale 11µm (0.1mil)

Scale 21μm: 0 - 1mm; 10μm: 1 - 5mm(0.1mil: 0 - 50mils; 1mil: 50 - 200mils)

Scale 1

±1-3% or ±2.5µm (±0.1mil)f

Scale 2

±1-3% or ±2.5µm (±0.1mil)f

Part Number

Probe Design

Minimum
Headroom

Minimum
Sample Diameter

T456CF1U

Ferrous Scale 1

85mm (3.35")

4mm (0.16")

T456CF1UARM

Ferrous Scale 1 Armoured Cable

139mm (5.47")

4mm (0.16")

T456CFNF1U

Dual FNF Scale 1

88mm (3.46")

Fg: 4mm (0.16")
Ng: 6mm (0.24")

T456CF2U

Ferrous Scale 2

89mm (3.50")

4mm (0.16")

Whichever is the greater. When calibrated using a sample of the uncoated base.
F: FNF probe in F mode; N: FNF probe in N mode.

www.elcometer.com

en-4

R

4 SPARES
Probe caps wear over time and the gauge will inform the user during
the calibration procedure if the cap needs replacingh. When tested
using smooth surfaces, probe caps have been scanned in excess of
50km (30 miles). Replacement caps are available to purchase fromElcometer or your local supplier.
Description
Scan Probe Caps (3 per pack)
Foil Set; Scale 1; 0 - 1500µm (0 - 60mils)
Foil Set; Scale 1; 0 - 1500µm (0 - 60mils) - Certified
Foil Set; Scale 2; 0 - 5mm (0 - 200mils)
Foil Set; Scale 2; 0 - 5mm (0 - 200mils) - Certified
Ferrous Zero Test Plate, 1%
Non-Ferrous Zero Test Plate, 1%Part Number
T456C23956
T99022255-1
T99022255-1C
T99022255-2
T99022255-2C
T9994910T9994911-

5 IMPORTANT INFORMATION“Scan” and “Auto Repeat” modes are only available on Elcometer 456
Model T separate gauges. The probe cap must be fitted to the Scan
Probe when used in these modes to avoid damage to the probe tip.
Use in these modes without the cap will invalidate the probe warranty.
Scan Probes can be used in “Standard” mode - with or without theprobe cap fitted - when connected to any Elcometer 456 separate
gauge.
When fitted with the probe cap, Scan Probes are not recommended foruse on coating thicknesses below 100µm (4mils).

hWhen the cap thickness falls below 200µm (8mils).

en-5

www.elcometer.com

R

www.elcometer.com

en-6

R

TMA-0568 Issue 04 - Text with cover 23959

R

Guide d'utilisation
Elcometer 456
Sondes Scanwww.elcometer.com

R

SOMMAIRE
SectionPrésentation
1
2
Calibrer avec une sonde scan
3
Caractéristiques techniques
4
Pièces de rechange
5
Information importante

Page
en-2
en-3
en-4
en-5
en-5

En cas de doute, merci de vous référer à la version originale anglaise de ce manuel.
Ce produit est livré dans un emballage en carton. Merci d'éliminer tous les emballages de manière écologique.
Contactez la déchèterie de votre localité pour plus d'informations sur le recyclage.
est une marque déposée d'Elcometer Limited, Edge Lane, Manchester, M43 6BU. Royaume Uni.
All other trademarks acknowledged.
© Elcometer Limited 2008 - 2018. Tous droits réservés. Aucune partie de ce document ne peut être reproduite,transmise, transcrite, stockée (dans un système documentaire ou autre) ou traduite dans quelque langue que cesoit, sous quelque forme que ce soit ou par n'importe quel moyen (électronique, mécanique, magnétique, optique,manuel ou autre) sans la permission écrite préalable d’Elcometer Limited.
R

fr-1

www.elcometer.com

R

1 PRÉSENTATIONLes Sondes Scan pour Elcometer 456 ont été spécialement conçuespour fonctionner avec les modes "Scan" et "Auto Repeat" disponibles
sur les jauges "Elcometer 456 Top sonde séparée". Elles sont livrées
avec un capuchon de sondea amovible, très résistant et remplaçable
par l'utilisateur. Il permet de prendre des mesures individuelles ou descanner de larges surfaces sans endommager la sonde ni lerevêtement.
Scan Mode : lorsque ce mode est
sélectionné, l'utilisateur peut faire glisser
la sonde Scan sur toute la surface. Quandil retire la sonde de la surface, la jauge
affiche la valeur moyenne d'épaisseur derevêtement (x), ainsi que la valeur maxi(Hi) et la valeur mini (Lo). Chaque sériede trois mesures peut apparaitre sur ungraphique et être enregistrée. Cette
fonction est idéale pour les inspections enligne.

Hi, x & Lo

1, 2, .......nMode Auto Repeat : lorsque ce mode est
sélectionné et que la sonde est placée surle substrat revêtu, la jauge prend
automatiquement des mesures à une
cadence de 140 par minute, jusqu'à ce
que l'utilisateur retire la sonde de lasurface. Chaque mesure prise est stockéedans la mémoire de l'instrument. Ce mode
permet d'accélérer sensiblement le travaild'inspection sur de larges surfaces tellesque poutrelles en I (RSJ's), citernes,
coques de navires, etc.
Note : "Mode Scan + maintenir" permet de relever la sonde de la surface pendant 1,5
secondes, et de la replacer sans débuter une nouvelle série de mesures.
L'utilisateur peut ainsi éviter les trous, joints de soudure, etc..situés sur lesubstrat sans affecter les mesures.

a

Des capuchons de rechange sont disponibles - voir détails Section 4 "Pièces de rechange" en page fr-4.

www.elcometer.com

fr-2

R

2 CALIBRER AVEC UNE SONDE SCAN
Lorsque vous utilisez une sonde Scan en mode "Scan" ou "AutoRepeat"b, vous devez installer le capuchon de sonde pour éviter
d'endommager la sonde elle-même. Pour calibrer, vous devez utiliser
exclusivement la calibration sur surface "Lisse" ou "Rugueuse/2 points".
Echelle 1 : Pour calibrer sur surface "Lisse", il faut une feuille de1500 µm (60 mils), et un substrat non revêtu ou une plaque zéro. Pourcalibrer sur surface "Rugueuse/2 points", il faut en plus une feuille de50 µm (2 mils).
Echelle 2 : Pour calibrer sur surface "Lisse", il faut une feuille de 4 mm
(160 mils)c, et un substrat non revêtu ou une plaque zéro. Pour calibrer
sur surface "Rugueuse/2 points", il faut en plus une feuille de50 µm (2 mils).
2.1 CONNECTER LA SONDE SCAN
1 Tournez le connecteur de la sondepour aligner les broches.
2 Vissez sur la colerette dans le sens
des aiguilles d'une montre.

21

2.2 CALIBRER EN MODE “SCAN”
OU “AUTO REPEAT”b:
1 Appuyez sur la touche Cal etsélectionnez "Méthode Cal”.
2 Sélectionnez "Lisse" ou "Rugueuse/2 points" suivi de "Calibrer" d.
3 Suivez les instructions à l'écran.
Pour éviter d'endommager la pointe de la sonde en mode "Scan" ou"Auto Repeat", il faut utiliser un embout de sonde (capuchon). Pendant lacalibration, lorsque la jauge le demande, si aucun embout n'est installé,la jauge se remet automatiquement en mode de mesure "Standard".
Les sondes Scan utilisent la fonction d'offset brevetée de l'Elcometer
456 garantissant que la moindre usure du capuchon dûe à l'utilisation
est prise en compte pendant la procédure de calibration. A la fin de laprocédure, la plage de mesure de la sonde et la combinaison ducapuchone sont affichées.
bPour sélectionner le mode de mesure, allez dans Menu/Initialiser/Mode Mesures.
La sonde Echelle 2 est fournie avec des feuilles d'étalonnage (câles) de 3mm (120mils) et 1mm (40mils),
qui peuvent être utilisées pour former une feuille d'étalonnage de 4mm (160mils).
d
Si vous avez choisi une autre méthode de calibration, la jauge vous demande de sélectionner "Lisse" ou"Rugueuse/2 points”.
e
Les capuchons de sondes ont une épaisseur nominale de 500 µm (20 mils).
c

fr-3

www.elcometer.com

R

3 CARACTÉRISTIQUES TECHNIQUES
3.1 AVEC CAPUCHON DE SONDE
Plage

Résolution

Précision

Echelle 1

100 - 1000µm (4 - 40mils)

Echelle 20,1 - 4,5mm (4 - 180mils)

Echelle 1

1µm (0,1mil)

Echelle 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0,1mil: 0 - 50mils; 1mil: 50 - 200mils)

Echelle 1±3% ou ±2,5µm (±0,1mil)f

Echelle 2±1-3% ou ±0,02mm (±1mil)f

Référence article Modèle de sonde

Hauteurminimale requiseDiamètre
minimum requis

T456CF1U

Ferreuse; Echelle 1

86mm (3,38")15mm (0,59")

T456CF1UARM

Ferreuse; Echelle 1- câble blindé

140mm (5,51")

15mm (0,59")

T456CFNF1U

Dual FNF; Echelle 1

89mm (3,50")

15mm (0,59")

T456CF2U

Ferreuse; Echelle 2

90mm (3,54")15mm (0,59")

3.2 SANS CAPUCHON DE SONDE
Plage

Résolution

Précision

Echelle 1

0 - 1500µm (0 - 60mils)

Echelle 2

0 - 5mm (0 - 200mils)

Echelle 1

1µm (0,1mil)

Echelle 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0,1mil: 0 - 50mils; 1mil: 50 - 200mils)

Echelle 1±1-3% ou ±2,5µm (±0,1mil)f

Echelle 2±1-3% ou ±2,5µm (±0,1mil)f

Référence article Modèle de sondefgHauteurminimale requiseDiamètre
minimum requis

T456CF1U

Ferreuse; Echelle 1

85mm (3,35")

4mm (0,16")

T456CF1UARM

Ferreuse; Echelle 1- câble blindé

139mm (5,47")

4mm (0,16")

T456CFNF1U

Dual FNF; Echelle 1

88mm (3,46")

Fg: 4mm (0,16")
Ng: 6mm (0,24")

T456CF2U

Ferreuse; Echelle 2

89mm (3,50")

4mm (0,16")

Selon le plus élevé. Calibration à l'aide d'un échantillon de substrat non revêtu.
F: sonde FNF en mode F; N: sonde FNF en mode N.

www.elcometer.com

fr-4

R

4 PIÈCES DE RECHANGE
Les capuchons de sondes s'usent avec le temps. Si le capuchon doitêtre remplacéh, la jauge en informe l'utilisateur pendant la procédure
de calibration. Lors de tests sur surfaces lisses, les capuchons ont
supporté des scans sur plus de 50 Km (30 miles). Vous pouvez vous
procurer des capuchons de rechange chez Elcometer ou votre
fournisseur local.
Description
Référence article
Capuchons pour sonde Scan (lot de 3)
T456C23956
Jeu de cales; Echelle 1; 0 - 1500µm (0 - 60mils)
T99022255-1
Jeu de cales; Echelle 1; 0 - 1500µm (0 - 60mils) - Certifié T99022255-1C
Jeu de cales; Echelle 2; 0 - 5mm (0 - 200mils)
T99022255-2
Jeu de cales; Echelle 2; 0 - 5mm (0 - 200mils) - Certifié T99022255-2C
Plaque zéro ferreuse, 1%
T9994910Plaque zéro non ferreuse, 1%
T99949115 INFORMATION IMPORTANTE
Les modes "Scan" et "Auto Repeat" ne sont disponibles que sur les"Elcometer 456 Top - Sonde séparée". Lorsque ces modes sontutilisés, le capuchon doit être positionné sur la sonde Scan pour éviter
d'endommager la pointe de la sonde. La garantie sera invalidée sivous utilisez ces modes de mesure sans capuchon.
Les sondes Scan peuvent être utilisées en mode "Standard" - ou sansle capuchon - lorsqu'elles sont connectées aux autres jauges
Elcometer 456 sonde séparée.
Une fois le capuchon de sonde installé, il est déconseillé d'utiliser lasonde Scan pour des épaisseurs de revêtement inférieures à 100 µm(4 mils).

h

Lorsque l'épaisseur du capuchon est inférieure à 200 µm (8 mils).

fr-5

www.elcometer.com

R

www.elcometer.com

fr-6

R

TMA-0568-01 Numéro 04 - Texte avec couverture N° 23959

R

Gebrauchsanleitung
Elcometer 456
Scan-Sonden

www.elcometer.com

R

INHALT
AbschnittÜberblick
1
2
Kalibrieren mit einer Scan-Sonde
3
Technische Daten
4
Ersatzteile
5
Wichtige Informationen

Seite
de-2
de-3
de-4
de-5
de-5

Beziehen Sie sich im Zweifelsfall bitte auf die englischsprachige Version.
Dieses Produkt ist in einem Karton verpackt. Stellen Sie bitte sicher, dass alle Verpackung auf
umweltverträgliche Weise entsorgt wird. Lassen Sie sich von Ihrer örtlichen Umweltbehörde weiterberaten.
ist ein eingetragenes Warenzeichen der Elcometer Limited, Edge Lane, Manchester, M43 6BU,
Großbritannien und Nordirland.
Alle anderen Handelsmarken sind anerkannt.
© Elcometer Limited 2008 - 2018. Sämtliche Rechte vorbehalten. Kein Teil dieses Dokuments darf ohne dievorherige schriftliche Genehmigung der Elcometer Limited in jedweder Form oder auf jedwede Art reproduziert,übertragen, transkribiert, gespeichert (in einem Abrufsystem oder auf sonstige Weise) oder in jedwede Sprache(elektronisch, mechanisch, magnetisch, optisch, manuell oder auf sonstige Weise) übersetzt werden.
R

de-1

www.elcometer.com

R

1 ÜBERLICKDie Elcometer 456 Scan-Sonden sind speziell für den Scan- undWiederholautomatikmodus des separaten Elcometer 456
Messgerätemodells T konzipiert. Sie werden mit einer hochverschleißfesten, durch den Benutzer austauschbaren Sondenkappea mit'Schnappverschluss' geliefert und ermöglichen sowohl das Erfasseneinzelner Messwerte als auch das schnelle Scannen größerer Flächen,ohne die Sonde oder die Beschichtung zu beschädigen.
Scanmodus: Wenn der Scanmodus gewählt
ist, kann die Scan-Sonde über die gesamte
Oberfläche gefahren werden. Wenn die
Sonde von der Oberfläche abgehoben wird,
zeigt das Messgerät den durchschnittlichen
Beschichtungsdickenwert (x) zusammen mit
dem maximalen (Hi) und minimalen (Lo)Dickenmesswert an. Jede Gruppe von drei
Messwerten kann auf dem Verlaufsdiagramm
angezeigt und im Speicher abgelegt werden.
Dieser Modus ist ideal für die Inspektion in
der Fertigung geeignet.
Wiederholautomatik: Wenn die Sonde bei
Verwendung der Wiederholautomatik auf die
beschichtete Fläche aufgesetzt wird, erfasst
sie automatisch automatisch 140 Messwerte
pro Minute, bis sie wieder von der Fläche
abgehoben wird. Jeder einzelne Messwert
wird im Speicher abgelegt. Die
Wiederholautomatik kann eine erheblichschnellere Inspektion großer beschichteter
Flächen wie zum Beispiel Stahlträger,Tanks, Schiffsrümpfe usw. bewirken.

Hi, x & Lo

1, 2, .......nHinweis: Bei Verwendung des "Scanmodus mit Haltefunktion" kann die Sonde bis zu
1,5 Sekunden lang von der Fläche abgehoben und erneut aufgesetzt werden,
ohne einen neuen Satz von Messwerten anzufangen. Dies ermöglicht dem
Anwender das Vermeiden von Löchern, Schweißnähten usw. im Substrat,ohne die Messergebnisse zu beeinträchtigen.

aErsatzsondenkappen sind ebenfalls erhältlich - für weitere Details siehe Abschnitt 4 "Ersatzteile" auf Seite de-5.

www.elcometer.com

de-2

R

2 KALIBRIEREN MIT EINER SCAN-SONDEBei Verwendung einer Scan-Sonde im Scan- oder Wiederholautomatikmodusb
muss die Sondenkappe angebracht sein, um eine Beschädigung derSondenspitze zu vermeiden und das Messgerät kann nur mit der
Kalibriermethode “Glatt” oder “Rau/2-Punkt” kalibriert werden.
Skala 1: Für die Glattkalibrierung sind eine 1500μm (60mil) Kalibrierfolieund ein unbeschichteter Untergrund oder eine Nullplatte erforderlich. Fürdie Rau/2-Punktkalibrierung ist ebenfalls eine 50μm (2mil) Kalibrierfolie
erforderlich.
Skala 2: Für die Glattkalibrierung sind eine 4mm (160mil) Kalibrierfoliec und
ein unbeschichteter Untergrund oder eine Nullplatte erforderlich. Für die
Rau/2-Punktkalibrierung ist ebenfalls eine 50μm (2mil) Kalibrierfolie
erforderlich.
2.1 ANSCHLIESSEN DER SCAN-SONDE
1 Drehen Sie den Sondenstecker, um die
Kontakte aufeinander auszurichten.
2 Drehen Sie den Schraubverschluß im
Uhrzeigersinn ein.

21

2.2 KALIBRIERUNG IM SCAN- ODERbWIEDERHOLAUTOMATIKMODUS :1 Drücken Sie die Taste “Kal" und wählen Sie “Kal-Methode".
2 Wählen Sie “Glatt” oder “Rau/2-Punkt” und dann “Kalibrieren” d.
3 Folgen Sie den Anleitungen auf dem Bildschirm.
Um eine Beschädigung der Sondenspitze im Scan- oder
Wiederholautomatikmodus zu vermeiden, muss eine Sondenkappe
angebracht werden. Wenn während der Kalibrierung nach Aufforderung
durch das Messgerät keine Kappe verwendet wird, schaltet das Messgerät
automatisch in den Standardmessmodus um.
Die Scan-Sonde verwendet die patentierte Offset-Funktion des Elcometer
456, die gewährleistet, dass jeder gebrauchsbedingter Kappenverschleißbei der Kalibrierung berücksichtigt wird. Am Ende des Kalibriervorgangs
wird der maximale Beschichtungsdickenbereich für die Kombination vonSonde und Kappee angezeigt.
bWählen Sie den Messmodus mit Menü / Einstellung / Mess-Modus aus.
Die Sonde der Skala 2 wird geliefert mit einer 3mm (120mils) und einer 1mm (40mils) Folie aus denen
dann zusammen auch eine 4mm (160mils) Folie erzeugt werden kann.
dWenn eine andere Kalibriermethode gewählt wurde, fordert das Messgerät den Benutzer auf, "Glatt" oder
"Rau/2-Punkt" zu wählen.
eSondenkappen haben eine Nenndicke von 500µm (20 mil).
cde-3

www.elcometer.com

R

3 TECHNISCHE DATEN
3.1 SONDENKAPPE ANGEBRACHT
BereichAuflösungGenauigkeit

Skala 1100 - 1000µm (4 - 40mils)

Skala 20,1 - 4,5mm (4 - 180mils)

Skala 11µm (0,1mil)

Skala 21μm: 0 - 1mm; 10μm: 1 - 5mm(0,1mil: 0 - 50mils; 1mil: 50 - 200mils)

Skala 1

±3% oder ±2,5µm (±0,1mil)f

Skala 2

±1-3% oder ±0,02mm (±1mil)f

Bestellnummer

Sondenausführung

Minimaler
Arbeitsabstand

Minimaler Probendurchmesser

T456CF1U

Eisenmetall; Skala 1

86mm (3,38")15mm (0,59")

T456CF1UARM

Eisenmetall; Skala 1- Panzerkabel

140mm (5,51")

15mm (0,59")

T456CFNF1U

FNF-Kombi; Skala 1

89mm (3,50")

15mm (0,59")

T456CF2U

Eisenmetall; Skala 2

90mm (3,54")15mm (0,59")

3.2 OHNE SONDENKAPPE
BereichAuflösungGenauigkeit

Skala 1

0 - 1500µm (0 - 60mils)

Skala 20 - 5mm (0 - 200mils)

Skala 11µm (0.1mil)

Skala 21μm: 0 - 1mm; 10μm: 1 - 5mm(0,1mil: 0 - 50mils; 1mil: 50 - 200mils)

Skala 1

±1-3% oder ±2,5µm (±0,1mil)f

Skala 2

±1-3% oder ±0,02mm (±1mil)f

Bestellnummer

Sondenausführung

Minimaler
Arbeitsabstand

Minimaler Probendurchmesser

T456CF1U

Eisenmetall; Skala 1

85mm (3,35")

4mm (0,16")

T456CF1UARM

Eisenmetall; Skala 1- Panzerkabel

139mm (5,47")

4mm (0,16")

T456CFNF1U

FNF-Kombi; Skala 1

88mm (3,46")

Fg: 4mm (0,16")
Ng: 6mm (0,24")

T456CF2U

Eisenmetall; Skala 2

89mm (3,50")

4mm (0,16")f

Der jeweils größere Wert trifft zu. Beim Kalibrieren unter Verwendung einer Probe des unbeschichteten
Untergrunds.
g
F: FNF-Sonde im F-Modus ; N: FNF-Sonde im N-Modus .

www.elcometer.com

de-4

R

4 ERSATZTEILE
Sondenkappen weisen im Verlauf der Zeit Verschleißerscheinungenauf. Falls die Kappe ausgetauscht werden muss, weist das Messgerät
den Benutzer während der Kalibrierung darauf hinh. In Tests auf glatten
Oberflächen wurden mit Sondenkappen über 50 km (30 Meilen)
gescannt. Ersatzendkappen sind bei Elcometer oder Ihrem örtlichen
Händler erhältlich.
Beschreibung
Bestellnummer
Scan-Sondenkappen (3er-Pack)
T456C23956
Folienset; Skala 1; 0 - 1500µm (0 - 60mil)
T99022255-1
Folienset; Skala 1; 0 - 1500µm (0 - 60mil) - zertifiziert T99022255-1C
Folienset; Skala 2; 0 - 5mm (0 - 200mil)
T99022255-2
Folienset; Skala 2; 0 - 5mm (0 - 200mil) - zertifiziert T99022255-2C
Eisen-Nullplatte, 1%
T9994910Nichteisen-Nullplatte, 1%
T99949115 WICHTIGE INFORMATIONEN
Der Scan- und Wiederholautomatikmodus ist nur am separaten
Elcometer 456 Messgerätemodell T verfügbar. Bei der Verwendung
der Scan-Sonde in diesen Modi muss die Sondenkappe angebracht
sein, um eine Beschädigung der Sondenspitze zu vermeiden. Bei
einer Verwendung in diesen Modi ohne Kappe erlischt die
Sondengarantie.
Wenn die Scan-Sonde an ein beliebiges separates Elcometer 456
Messgerät angeschlossen ist, kann sie im Standardmodus mit oderohne angebrachter Sondenkappe verwendet werden.
Bei angebrachter Sondenkappe wird die Verwendung der ScanSonden für Beschichtungsdicken von weniger als 100µm (4mil) nicht
empfohlen.

hWenn die Kappendicke 200µm (8mil) unterschreitet.

de-5

www.elcometer.com

R

www.elcometer.com

de-6

R

TMA-0568-02 Ausgabe 04 - Text mit Umschlag Nr. 23959

R

Guía del usuario
Elcometer 456
Sondas Escan

www.elcometer.com

R

CONTENIDO
Sección
1
Descripción general2
Calibración utilizando una sonda escan
3
Especificaciones técnicas
4
Repuestos5
Informacion importante

Página
es-2
es-3
es-4
es-5
es-5

Para despejar cualquier duda, consulte la versión original en inglés.
Este producto se suministra en un paquete de cartón. Asegúrese de que este embalaje se desecha de forma
respetuosa con el medio ambiente. Consulte a las autoridades locales en materia medioambiental para obtener
información.
es una marca comercial registrada de Elcometer Limited, Edge Lane, Manchester, M43 6BU.
Reino Unido
Todas las demás marcas comerciales se dan por reconocidas.
© Elcometer Limited 2008 - 2018. Todos los derechos reservados. Este documento ni ningún fragmento del mismo
pueden reproducirse, transmitirse, transcribirse, almacenarse (en un sistema de recuperación o de otro tipo) ni
traducirse a ningún idioma, en ningún formato ni por ningún medio (ya sea electrónico, mecánico, magnético,óptico, manual o de otro tipo) sin permiso previo y por escrito de Elcometer Limited.
R

es-1

www.elcometer.com

R

1 DESCRIPCIÓN GENERAL
Las sondas Escan Elcometer 456 han sido diseñadas
específicamente para su uso con los modos "Escan" y "AutoRepetición" de los medidores separados Elcometer 456 Modelo T . Se
suministran con una tapa de la sondaa de gran durabilidad 'snap on' oajuste a presión, reemplazable por el usuario - que permite a losusuarios tomar las lecturas individuales o escanear rápidamente
grandes superficies sin dañar la sonda o el revestimiento.
Modo Escan: Cuando se selecciona elModo de Escan, los usuarios pueden
deslizar la sonda Escan a través de toda lasuperficie. A medida que la sonda se retirade la superficie el medidor muestra el valor
promedio de espesor de revestimiento (x)junto con el más alto (Hi) y más bajo (Lo)valores de espesor. Cada conjunto de tres
lecturas se pueden visualizar en un gráfico
de ejecución y se almacena en la memoria.
Esto es ideal para la inspección de línea.
Modo de Auto Repetición: Cuando lasonda se coloca sobre el sustrato
revestido en el modo de repetición
automática, el medidor toma lecturas
automáticamente a una velocidad de 140
por minuto hasta que la sonda se retira delsustrato. Cada lectura individual sealmacena en la memoria. El modo derepetición automática se puede utilizar
para acelerar significativamente el control
de grandes áreas revestidas como vigasde acero I- (RSJ), tanques, cascos debarcos, etc.

Hi, x & Lo

1, 2, .......nNota: "Modo Escan Retención" permite que la sonda sea levantada de la superficie por
un máximo de 1.5 segundos y vuelta a colocar sin necesidad de iniciar una
nueva serie de lecturas - que permite al usuario evitar agujeros, soldaduras, etc
en el sustrato, sin afectar los resultados de la medición.
aRepuestos de tapas de sonda están disponibles - ver Sección 4 "Recambios" en la página es-5 para más
detalles.

www.elcometer.com

es-2

R

2 CALIBRACIÓN UTILIZANDO UNA SONDA ESCAN
Cuando se utiliza una sonda Escan en modeb "Escan" o "AutoRepetición", la tapa de la sonda debe estar colocada para evitar daños
en la punta de la sonda y el medidor sólo puede ser calibrado utilizando
los métodos de calibración "Lisa" o "Rugosa / 2 Puntos".
Escala 1: Para calibración "Lisa" a 1500μm (60mils) lámina nominal ysin revestimiento se requiere una base / placa cero. Para el “Rugosa /2 Puntos" a 50μm (2mil) lámina nominal también es necesaria.
Escala 2: Para calibración "Lisa" a 4mm (160mils) lámina nominalc y sin
revestimiento se requiere una base / placa cero. Para el “Rugosa / 2
Puntos" a 50μm (2mil) lámina nominal también es necesaria.
2.1 CONECTAR LA SONDA ESCAN
1 Gire el enchufe de la sonda paraalinear las clavijas.
2 Atornillado en el cuello - En sentido
de las manecillas del reloj.
2.2 CALIBRACIÓN CUANDO ESTA EN
MODEb "ESCAN" O "AUTO
REPETICION":
1 Pulse la tecla Cal y seleccione"Método Cal”.
2 Seleccione "Lisa" o “Rugosa / 2
dPuntos", seguido de "Calibrar" .
3 Siga las instrucciones en pantalla.

2

1

Con el fin de evitar daños a la punta de la sonda en el modo "Escan" o"Auto Repetición", debe colocarse una tapa de sonda. Durante elprocedimiento de calibración si cuando se les pregunta por el medidor,
no es utilizada, el indicador volverá automáticamente al modo "estándar"de medición.
Las sondas Escan usan la característica patentada de compensacióndel Elcometer 456 asegurando que cualquier desgaste de la tapa
durante el uso está incorporado dentro del procedimiento de calibración,que muestra rango de espesor de revestimiento máximo de lae
combinación de la sonda y tapa al final de la rutina de calibración.
bPara seleccionar el modo de medida vaya a Menú/Ajustes/Modo de Medición.
La sonda Escala 2 se suministra con una lámina de 3mm (120mils) y otra de 1mm (40mils), las cuales sepueden utilizar para obtener una lámina de 4mm (160mm).
d
Si un método de calibración alternativo ha sido seleccionado, el medidor le pedirá al usuario elseleccionar "Lisa" o “Rugosa / 2 Puntos”.
eTapas de la sonda tiene un espesor nominal de 500μm (20 mils).
c

es-3

www.elcometer.com

R

3 ESPECIFICACION TECNICA
3.1 TAPA DE LA SONDA AJUSTADA
Rango

Resolución

Precisión

Escala 1

100 - 1000µm (4 - 40mils)

Escala 2

0.1 - 4.5mm (4 - 180mils)

Escala 1

1µm (0.1mil)

Escala 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0.1mil: 0 - 50mils; 1mil: 50 - 200mils)

Escala 1±3% o ±2.5µm (±0.1mil)f

Escala 2±1-3% o ±0.02mm (±1mil)f

Referencia

Diseño de la Sonda

Espacio libremínimoDiámetro Mínimo
de la Muestra

T456CF1U

Ferrosa; Escala 1

86mm (3.38")15mm (0.59")

T456CF1UARM

Ferrosa; Escala 1 cable blindado

140mm (5.51")

15mm (0.59")

T456CFNF1U

Dual FNF; Escala 1

89mm (3.50")

15mm (0.59")

T456CF2U

Ferrosa; Escala 2

90mm (3.54")15mm (0.59")

3.2 SIN LA TAPA DE LA SONDA
Rango

Resolución

PrecisiónfgEscala 1

0 - 1500µm (0 - 60mils)

Escala 2

0 - 5mm (0 - 200mils)

Escala 1

1µm (0.1mil)

Escala 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0.1mil: 0 - 50mils; 1mil: 50 - 200mils)

Escala 1±1-3% o ±2.5µm (±0.1mil)f

Escala 2±1-3% o ±2.5µm (±0.1mil)f

Referencia

Diseño de la Sonda

Espacio libremínimoDiámetro Mínimo
de la Muestra

T456CF1U

Ferrosa; Escala 1

85mm (3.35")

4mm (0.16")

T456CF1UARM

Ferrosa; Escala 1 cable blindado

139mm (5.47")

4mm (0.16")

T456CFNF1U

Dual FNF; Escala 1

88mm (3.46")

Fg: 4mm (0.16")
Ng: 6mm (0.24")

T456CF2U

Ferrosa; Escala 2

89mm (3.50")

4mm (0.16")

Cualquiera que sea el mayor. Cuando es calibrado utilizando una muestra de la base sin revestir.
F: sonda FNF en el modo F; N: sonda FNF en el modo N.

www.elcometer.com

es-4

R

4 REPUESTOS
Tapas de la sonda se desgastan con el tiempo y el medidor informará alusuario durante el proceso de calibración si la tapa necesita serhremplazada . Cuando se ha probado usando superficies lisas, las tapasde la sonda han sido escaneadas en exceso de 50 km (30 millas). Las
tapas de repuesto están disponibles para comprar de Elcometer o con
su proveedor local.
Descripción
Tapas de la Sonda Escan (3 por paquete)
Conjunto de láminas Escala 1; 0 - 1500µm (0 - 60mils)
Conjunto de láminas Escala 1; 0 - 1500µm (0 - 60mils) Certificadas
Conjunto de láminas Escala 2; 0 - 5mm (0 - 200mils)
Conjunto de láminas Escala 2; 0 - 5mm (0 - 200mils) Certificadas
Placa de Prueba cero Ferrosa, 1%
Placa de Prueba cero No-Ferrosa, 1%

Referencia
T456C23956
T99022255-1
T99022255-1C
T99022255-2
T99022255-2C
T9994910T9994911-

5 INFORMACION IMPORTANTE
Modos de "Escan" y "Repetición Automática" sólo están disponibles en
los medidores Elcometer 456 Modelo T separados. La tapa de la sonda
debe ser instalada en la sonda Escan cuando se usa en estos modos
para evitar daños en la punta de la sonda. El uso en estos modos sin latapa invalidará la garantía de la sonda.
Las sondas Escan se puede utilizar en modo "Estándar" - con o sin latapa de la sonda colocada - cuando se conecta a cualquier medidor
separado Elcometer 456.
Cuando las sondas Escan están equipadas con la tapa de la sonda, lassondas Escan no son recomendas para su uso en el revestimiento deespesores inferiores a 100μm (4mils).

hCuando el espesor de la tapa cae por debajo de 200μm (8mils).

es-5

www.elcometer.com

R

www.elcometer.com

es-6

R

TMA-0568-04 Edición 04 - Texto con cubierta 23959

R

Gebruikershandleiding
Elcometer 456
Scansondes

www.elcometer.com

R

INHOUDSOPGAVE
Sectie1Overzicht
2
Kalibreren van een scansonde
3
Technische specificaties
4
Reserveonderdelen
5
Belangrijke informatie

Pagina
nl-2
nl-3
nl-4
nl-5
nl-5

Raadpleeg de originele Engelse versie om twijfel uit te sluiten.
Dit product is verpakt in karton. Zorg ervoor dat alle verpakking milieuvriendelijk wordt afgevoerd. Neem contact
op met de milieuafdeling van uw gemeente voor advies.
is een gedeponeerd handelsmerk van Elcometer Limited, Edge Lane, Manchester, M43 6BU.
Verenigd Koninkrijk
Alle andere handelsmerken zijn het eigendom van hun respectievelijke eigenaars.
© Elcometer Limited 2008 - 2018. Alle rechten voorbehouden. Niets van dit document mag worden gereproduceerd,
overgedragen, getranscribeerd, opgeslagen (in een retrievalsysteem of anderszins) of vertaald in enige taal, in
enige vorm of door enig middel (elektronisch, mechanisch, magnetisch, optisch, handmatig of anderszins) zonderde voorafgaande schriftelijke toestemming van Elcometer Limited.
Rnl-1

www.elcometer.com

R

1 OVERZICHT
Elcometer 456 Scansondes zijn speciaal ontworpen voor gebruik met
de Scan- en Automatische herhaalmodi van Elcometer Model T metersvoor afzonderlijke sondes. Deze sondes zijn voorzien van een
duurzame, verwisselbare 'opzet' sondekapa waarmee u individuele
metingen kunt verrichten of snel grote oppervlakken kunt scannen
zonder de sonde of coating te beschadigen.
Scanmodus: Als u de scanmodus
selecteert, kunt u met de scansonde overhet gehele oppervlak schuiven. Als u desonde weer van het oppervlak afneemt,
toont de meter de gemiddelde (x), hoogste(Hi) en laagste (Lo) laagdiktewaarden. Elke
set van drie waarden kan worden
weergegeven in de grafiek en opgeslagenin het geheugen. Dit is ideaal voor online
inspecties.
Automatische herhaalmodus: Als u desonde in automatische herhaalmodus op
een gecoat substraat plaatst, verricht demeter automatisch 140 metingen per
minuut totdat u de sonde van het substraat
haalt. Elke individuele meting wordt
opgeslagen in het geheugen. De
Automatische herhaalmodus kunt u
gebruiken om de inspectie van grote
gecoate oppervlakken zoals stalen
H-profielen, tanks, scheepsrompen enz.
aanzienlijk te versnellen.

Hi, x & Lo

1, 2, .......nOpmerking: In 'Scanmodus met vasthoudfunctie' kunt u de sonde maximaal
1,5 seconden van het oppervlak afhalen en herplaatsen zonder dat er een
nieuwe set metingen wordt gestart. Hierdoor kunt u gaatjes, lasnaden enz.
in het substraat ontwijken zonder de meetresultaten te beïnvloeden.

aEr zijn reserve kappen beschikbaar voor de sondes – zie sectie 4 'Reserveonderdelen' op pagina nl-5
voor meer informatie.

www.elcometer.com

nl-2

R

2 KALIBREREN VAN EEN SCANSONDE
Als u een scansonde gebruikt in de Scan- of Automatischeherhaalmodusb moet de kap zijn aangebracht om te voorkomen dat desondekop beschadigt. U kunt de meter alleen kalibreren met dekalibratiemethoden 'Glad' of ‘Ruw/2-punts'.
Schaal 1: Voor de kalibratiemethode 'Glad' is een nominale folie van
1500µm (60mils) en een ongecoate basis-/nulplaat nodig. Voor dekalibratiemethode ‘Ruw/2-punts' is ook een nominale folie van
50µm (2mil) nodig.
Schaal 2: Voor de kalibratiemethode 'Glad' is een nominale folie van
4mm (160mils)c en een ongecoate basis-/nulplaat nodig. Voor dekalibratiemethode ‘Ruw/2-punts' is ook een nominale folie van
50µm (2mil) nodig.
2.1 DE SCANSONDE AANSLUITEN
1 Draai de sondeplug zodat depinnetjes gelijkstaan met decontracontacten van de meter. Steekde plug in de meter.
2 Borg de plug door de ring met deklok mee te draaien.

2

1

b

2.2 KALIBREREN IN SCAN- OF AUTOMATISCHE HERHAALMODUS :1 Druk op de softkey Kal en kies 'Kal Methode’.
d2 Selecteer ‘Glad' of ‘Ruw/2-punts' gevolgd door 'Kalibreren' .
3 Volg de instructies op het scherm.
U dient een sondekap aan te brengen om te voorkomen dat de sondekop
beschadigt tijdens gebruik van de Scan- of Automatische herhaalmodus.
Als de meter tijdens het kalibratieproces aangeeft dat u geen kap moet
gebruiken, zal de meter automatisch terugkeren naar de 'Standaard'
meetmodus.
Scansondes maken gebruik van de gepatenteerde correctiefunctie vande Elcometer 456. Hierdoor wordt slijtage aan de kap meegenomen inhet kalibratieproces en toont de meter uiteindelijk het maximale
laagdiktebereik van de sonde plus de sondekape.
b

Om de meetmodus te selecteren gaat u naar Menu/Setup/Meet modus.
The schaal 2 sonde wordt geleverd met een 3mm (120 mils) en 1mm (40mils) folie die gebruikt kunnen
worden om een 4mm (160mils) folie te maken (op elkaar leggen).
d
Als er een alternatieve kalibreermethode is geselecteerd zal de meter aangeven om 'Glad' of 'Ruw/
2-punts' te kiezen.
eKappen van sondes hebben een nominale dikte van 500µm (20mils).
c

nl-3

www.elcometer.com

R

3 TECHNISCHE SPECIFICATIES
3.1 MET KAP OP SONDE
BereikResolutie

Nauwkeurigheid

Schaal 1

100 - 1000µm (4 - 40mils)

Schaal 20,1 - 4,5mm (4 - 180mils)

Schaal 1

1µm (0,1mil)

Schaal 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0,1mil: 0 - 50mils; 1mil: 50 - 200mils)

Schaal 1±3% of ±2,5µm (±0,1mil)f

Schaal 2±1-3% of ±0,02mm (±1mil) (±1mil)f

Onderdeelnummer

Sondeontwerp

Minimalekoplengte

Minimalemonsterdiameter

T456CF1U

Ferro; Schaal 186mm (3,38")15mm (0,59")

T456CF1UARM

Ferro; Schaal 1 gepantserde kabel

150mm (5,51")

15mm (0,59")

T456CFNF1U

Dual FNF; Schaal 1

89mm (3,50")

15mm (0,59")

T456CF2U

Ferro; Schaal 2

90mm (3,54")15mm (0,59")

3.2 ZONDER KAP OP SONDE
BereikResolutie

NauwkeurigheidfgSchaal 1

0 - 1500µm (0 - 60mils)

Schaal 2

0 - 5mm (0 - 200mils)

Schaal 1

1µm (0.1mil)

Schaal 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0,1mil: 0 - 50mils; 1mil: 50 - 200mils)

Schaal 1±1-3% of ±2,5µm (±0,1mil)f

Schaal 2±1-3% of ±0,02mm (±1mil)f

Onderdeelnummer

Sondeontwerp

Minimalekoplengte

Minimalemonsterdiameter

T456CF1U

Ferro; Schaal 1

85mm (3,35")

4mm (0,16")

T456CF1UARM

Ferro; Schaal 1 gepantserde kabel

159mm (5,47")

4mm (0,16")

T456CFNF1U

Dual FNF; Schaal 1

88mm (3,46")

Fg: 4mm (0,16")
Ng: 6mm (0,24")

T456CF2U

Ferro; Schaal 2

89mm (3,50")

4mm (0,16")

Welke er groter is. Als er gekalibreerd is met een monster van de ongecoate basis.
F: FNF sonde in F-Modus; N: FNF sonde in N-Modus

www.elcometer.com

nl-4

R

4 RESERVEONDERDELEN
Sonde-kappen slijten na verloop van tijd en de meter informeert utijdens het kalibratieproces of de kap vervangen dient te wordenh. Bij
gladde oppervlakken werd met de sonde-kappen meer dan 50 km (30mijl) gescand. Reservekappen kunt u aanschaffen via Elcometer of uw
lokale leverancier.
Beschrijving
Kappen voor scansondes (3 stuks per pakket)
Folieset; schaal 1, 0 - 1500µm (0 - 60mils)
Folieset; schaal 1, 0 - 1500µm (0 - 60mils) gecertificeerd
Folieset; schaal 2, 0 - 5mm (0 - 200mils)
Folieset; schaal 2, 0 - 5mm (0 - 200mils) gecertificeerd
Ferro nulplaat, 1%
Non-ferro nulplaat, 1%

Onderdeelnummer
T456C23956
T99022255-1
T99022255-1C
T99022255-2
T99022255-2C
T9994910T9994911-

5 BELANGRIJKE INFORMATIE
De Scan- en Automatische herhaalmodi zijn alleen beschikbaar bij
Elcometer 456 Model T meters voor afzonderlijke sondes. U dient een
sondekap aan te brengen om te voorkomen dat de sondekop
beschadigt tijdens gebruik van de scansondes in deze modi. Bij
gebruik van sondes zonder kap in deze modi komt de garantie tevervallen.
De Scansondes kunt u ook in 'Standaardmodus' gebruiken – met ofzonder kap – met elke Elcometer 456 meter voor afzonderlijke sondes.
Voor Scansondes met eindkappen wordt gebruik niet aangeraden voor
laagdiktes onder de 100µm (4mils).

h

Als de kapdikte onder de 200µm (8mils) valt.

nl-5

www.elcometer.com

R

www.elcometer.com

nl-6

R

TMA-0568-11 Uitgave 04 - Text met voorblad No. 23959

R

用户手册
Elcometer 456
扫描www.elcometer.com

头

R

目录
分

页

1

览

2

使用扫描

3

技术规格

zh-4

4

备件

zh-5

5

重要

疑议, 请
该产品是
R

有

zh-2

zh-5

息

英

本.

在一个纸箱

.请

是Elcometer

司的注册

标也都得

zh-3

头时的

保

有

环

感的

式处理. 请

询当地环

局为进一

指导.

标, Edge Lane, 曼彻斯 , M43 6BU,英国.

注册许可.

© Elcometer Limited 2008 - 2018. 司保有权利. 本任何 分都不得复制, 传输, 存储(在检 或其他)，或者
在 有Elcometer Limited事先书面许可的情况
任何 式(电子, 机械, 性, 光学, 手动或其他)译成任何语 .

zh-1

www.elcometer.com

R

1 览
易 456扫描 头是为配合456 级分体型测厚仪上的“扫描”和“自动重复”模式的使用而专门设计的. 头配有经久耐用的保护 a，使用者可进行更换， 头保护 的作用——使用者可获 单点读 或在大面积 面
上进行快速扫描，而不损坏 头或涂层.
扫描模式：当选 扫描模式，使用者可在整个 面 动扫描 头. 头 开 面
时，仪 显示涂层厚 平 值x,
大值Hi和 小值Lo. 上述三个读可显示在曲线图上，并存
存. 这对线性检测常 用.

Hi, x & Lo

自动重复模式：当选 自动重复模式，头在 面上移开 前，仪
分钟140读 的速 获 厚 值.
个读存存. 自动重复模式可大大加快I型钢 ，大， 坞 的测
厚速 .
注: 扫描锁 模式 ，可 获 一组连续读 ，其间 许
使用者可 开小洞、焊缝等，而不 响测量结果.

a可获得头盖配件——详见

www.elcometer.com

分的

1, 2, .......n

头移开

面

长1.5

. 这使得

三页“备件”

zh-2

R

2 使用扫描 头时的
使用扫描 头时的”扫描”或”自动重复”模式 b， 头盖必须盖上，
头 点损坏，仪 仅 进行“平 ”
或“粗 /两点”.
量程 1: 平 ”
需1500µm（60mils）校准膜片和 涂层基体或粗 /两点”
还需要50µm（2mils）膜片.
量程 2: 平 ”
需4mmc（160mils）校准膜片和 涂层基体或粗 /两点”
还需要50µm（2mils）膜片.
2.1 连 扫描 头
1
转 头 对
口.
2 顺时针 紧 钮.
2.2 “扫描”或“自动重复”模式 的1
软键，选择“”.
2 选择“平 ”或“粗 /两点”
，
d
再 “
”.
3
照屏幕上的指示.

.

.

2

b

:

1

为
“扫描”或“自动重复”模式头 点的损坏，需要 好 头保护.
过程中，仪提示 好保护 ，若不进行这一操作，仪
自动转 “标 ”测量模式.
扫描 头使用了易 456的专利offset功 ， 保使用过程中任何时候保护进
程，在程结束前显示 头和保护 e组合的大厚 测试范围.

b

要选测量模式， 菜单/设 /测量模式进 .
量程2 头提供 3mm(120mil) 和 1mm(40 mil) 膜片,可 用来做 4mm(160mils) 膜片.
d
若选择了其他的，仪
提示使用者选择“平 ”或“粗 /两点”.
e头盖的名义厚 为500µm (20mils).
czh-3

www.elcometer.com

R

3 技术规格3.1 好了 头保护
范围

分

精

量程 1

100 - 1000µm (4 - 40mils)

量程 2

0.1 - 4.5mm (4 - 180mils)

量程 1

1µm (0.1mil)

量程 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0.1mil: 0 - 50mils; 1mil: 50 - 200mils)

量程 1±3% 或 ±2.5µm (±0.1mil)f

量程 2±1-3% 或 ±0.02mm (±1mil)f

型

头设计T456CF1U

铁基F; 量程 1

T456CF1UARM

铁基F; 量程 1-铠

15mm (0.59")线

150mm (5.51")

15mm (0.59")

T456CFNF1U

铁基 铁基两用FNF;
量程 1

89mm (3.50")

15mm (0.59")

T456CF2U

铁基F; 量程 290mm (3.54")15mm (0.59")

好

头保护

范围

分

精

量程 1

0 - 1500µm (0 - 60mils)

量程 2

0 - 5mm (0 - 200mils)

量程 1

1µm (0.1mil)

量程 2

1μm: 0 - 1mm; 10μm: 1 - 5mm(0.1mil: 0 - 50mils; 1mil: 50 - 200mils)

量程 1±1-3% 或 ±2.5µm (±0.1mil)f

量程 2±1-3% 或 ±2.5µm (±0.1mil)f

型

g

小样

86mm (3.38")

3.2 未

f

小净

头设计

T456CF1U

铁基F; 量程 1

T456CF1UARM

铁基F; 量程 1-铠

小净小样

85mm (3.35")

4mm (0.16")149mm (5.47")

4mm (0.16")

T456CFNF1U

铁基 铁基两用FNF;
量程 1

88mm (3.46")

Fg: 4mm (0.16")
Ng: 6mm (0.24")

T456CF2U

铁基F; 量程 289mm (3.50")

4mm (0.16")

较大者. 当用 涂层样
F: FNF探头的F模式; N: FNF

www.elcometer.com

线

时.
头的N模式

zh-4

R

4 备件
头保护
损，当保护 需要更换时，仪在过程中进行提h示 . 当平面的测量长 超过50km（30英里），探头保护 需要更换，可向易 或当地经销 购买.
描述扫描 头保护 （3只 ）膜片组；量程1；0 - 1500µm (0 - 60mils)
膜片组；量程1；0 - 1500µm (0 - 60mils)-带证书
膜片组；量程2；0 - 5mm (0 - 200mils)
膜片组；量程2；0 - 5mm (0 - 200mils)-带证书
铁基
，1%
铁基
，1%

件编
T456C23956
T99022255-1
T99022255-1C
T99022255-2
T99022255-2C
T9994910T9994911-

5 重要 息“扫描”和“自动重复”模式仅存在于456 级型分体测厚仪. 扫描 头需
保护 配合使用，
头 点损坏. 这两个模式 ，若不使用保护，仪 不在保 范围 .
当连在任何456主机上，扫描
使用保护
可.
当

h

上保护

当保护

zh-5

厚

，不

议在

头可在“标

”测量模式

使用，使用或不于100µm (4mils)的涂层上使用扫描头.

小于200µm (8mils)时.

www.elcometer.com

R

www.elcometer.com

zh-6

R

TMA-0568-10

04版 - 有盖本23959R

ユーザーガイド
Elcometer 456
スキャンプローブ

www.elcometer.com

R

⽬次
ページ

セクション
1

概要

jp-2

2

スキャンプローブを使った校正

jp-3

3

仕様

jp-4

4

交換部品

jp-5

5

注意事項

jp-5

不明な点がある場合は、英語版の取扱説明書を確認してください。
本製品は、段ボール箱に梱包されています。包材は、環境保全に配慮した⽅法で破棄してください。詳しくは、
地⽅⾃治体等の適切な機関にお問い合わせください。
は、Elcometer Limitedの登録商標です。所在地：Edge Lane, Manchester, M43 6BU
United Kingdom
その他の商標については、その旨が記されています。
R© Elcometer Limited 2008 - 2018. この⽂書の⼀部または全部を、Elcometer Limitedの事前の書⾯による許可なく、いかなる形式や⽅法（電⼦的、機械的、磁気的、⼯学的、⼿動を問わず）によっても、複製、転送、保管（検索可能なシステムかどうかを問わず）、または他の⾔語に翻訳することを禁じます。

jp-1

www.elcometer.com

R

1 概要
Elcometer 456スキャンプローブは、Elcometer 456モデルTセパレー
ト型膜厚計をScan（スキャン）モードとAuto Repeat（⾃動測定）モ
ードで使⽤する場合に必要になります。耐久性が⾼く簡単に着脱可能なキャップaが付いており、個々の膜厚値を読み取ったり、広い⾯をプ
ローブや塗膜を傷つけずに素早くスキャンできます。
スキャンモード：スキャンプローブで試験⾯をなぞるようにして測定することができます。試験⾯からプローブを離した
ときに、読み取った最⼤膜厚（Hi）と最⼩膜厚（Lo）、平均膜厚（x）が画⾯に表⽰されます。これらの値の時間の経過に伴う変化をグラフで表⽰したり、メモリに保存したりできます。オンライン検査に適しています。
⾃動測定モード：下地の上にプローブを置くだけで、毎分140回の速度で⾃動的に膜厚を読み取ります。読み取り値はメモリに保存されます。⾼速で読み取るので、（RSJの）鋼Iビームやタンク、船舶の胴体などの広い⾯の測定に最適です。

Hi, x & Lo

1, 2, .......n

注：スキャンモードでホールド機能を使うと、プローブを試験⾯から1.5秒間離して、新しい読み取りを開始することなく、プローブを交換できます。下地に⽳を開けたり、溶接継⽬を付けたりする⼼配がなく、検査結果にも影響しません。

a

プローブの交換⽤キャップについては、セクション4の「交換⽤部品」（jp-5ページ）を参照してください。

www.elcometer.com

jp-2

R

2 スキャンプローブを使った校正
スキャンモードまたは⾃動測定モードbでスキャンプローブを使⽤するときは、プローブの先端を傷つけないようにキャップを装着し、膜厚計の校正⽅法には必ず平滑⾯または粗⾯/2ポイントを使⽤してください。
スケール1:平滑⾯で校正する場合は、1500µm（60mil）の標準フォイ
ルと、塗装前の素地またはゼロ点調整板が必要になります。粗⾯/
2ポイントの場合も、50µm（2mil）の標準フォイルが必要になります。
スケール2:平滑⾯で校正する場合は、4mmc（160mil）の標準フォイ
ルと、塗装前の素地またはゼロ点調整板が必要になります。粗⾯/
2ポイントの場合も、50µm（2mil）の標準フォイルが必要になります。
2.1 スキャンプローブの接続
1 プローブのプラグを本体端⼦のピンの並びに合わせます。
2 プローブを差し込んで右に回します。
2.2 スキャンモードまたは⾃動測定
モードbでの校正:
1 調整ソフトキーを押して、調整⽅法を選択します。
2 平滑⾯または粗⾯/2ポイントを選択して、調整するを選択します。d
3 画⾯に表⽰される指⽰に従います。

2

1

スキャンモードまたは⾃動測定モードでの使⽤中は、プローブの先端を傷つけないようにキャップを装着してください。校正中にキャップを装着しないようにという指⽰が画⾯に表⽰される場合は、膜厚計が⾃動的に標準モードに戻ります。
スキャンプローブではゼロ点移動機能（特許取得済み）を使⽤できるので、校正の最後の段階で表⽰されるプローブの最⼤膜厚範囲とキャップeの組み合わせを⾒ながら摩耗量を考慮できます。
測定モードを選択するには、メニュー/設定/測定モードの順に選択します。
スケール２のプローブには、3mm(120mils)と1mm(40mils)のフォイルが付属し、重ねると4mm(160mils)で使⽤できます。
d
校正⽅法を選択すると、Smooth (平滑⾯）Rough/ 2 Point (粗⾯/2ポイント）のどちらかを選択するように指⽰が表⽰されます。
e
プローブのキャップの厚さは、通常500µm（20mil）です。
b

c

jp-3

www.elcometer.com

R

3 仕様
3.1 キャップ装着時
測定範囲

分解能

精度

スケール1

100～1000µm (4～40mils)

スケール2

0.1～4.5mm (4～180mils)

スケール11µm (0.1mil)

スケール21μm: 0～1mm; 10μm: 1～5mm(0.1mil: 0～50mils; 1mil: 50～200mils)

スケール1±3% または ±2.5µm (±0.1mil)

スケール2±1-3% または ±0.02mm (±1mil)

f
f

天地余裕
（最⼩）

最⼩測定域直径

磁性⾦属⽤; スケール186mm (3.38")15mm (0.59")

T456CF1UARM

磁性⾦属⽤; スケール1⾦属補強ケーブル付き

140mm (5.51")

15mm (0.59")

T456CFNF1U

磁性·⾮磁性⾦属両
⽤;スケール1

89mm (3.50")

15mm (0.59")

T456CF2U

磁性⾦属⽤; スケール2

90mm (3.54")15mm (0.59")

スケール10～1500µm (0～60mils)

スケール20～5mm (0～200mils)

スケール11µm (0.1mil)

スケール21μm: 0～1mm; 10μm: 1～5mm(0.1mil: 0～50mils; 1mil: 50～200mils)

スケール1±1-3% または ±2.5µm (±0.1mil)

スケール2±1-3% または ±0.02mm (±1mil)

コード番号

プローブの型

T456CF1U

3.2 キャップなし
測定範囲

分解能

精度

f
g

f

f

天地余裕
（最⼩）

最⼩測定域直径

磁性⾦属⽤; スケール185mm (3.35")

4mm (0.16")

T456CF1UARM

磁性⾦属⽤; スケール1⾦属補強ケーブル付き

139mm (5.47")

4mm (0.16")

T456CFNF1U

磁性·⾮磁性⾦属両
⽤;スケール1

88mm (3.46")

Fg: 4mm (0.16")
Ng: 6mm (0.24")

T456CF2U

磁性⾦属⽤; スケール2

89mm (3.50")

4mm (0.16")

コード番号

プローブの型

T456CF1U

いずれか⼤きい⽅塗装前の素地を使って校正した場合.
F: FNFプローブをFモードで使⽤; N: FNFプローブをNモードで使⽤

www.elcometer.com

jp-4

R

4 交換部品
プローブのキャップは何度も使⽤しているうちに摩耗します。交換が必要になると、校正中にそのことが画⾯に表⽰されますh。キャップが摩耗するまで滑らかな⾯を50km（約30マイル）スキャンできます。
交換⽤のキャップは、Elcometerまたは最寄りの代理店からお求めいただけます。
説明
コード番号
スキャンプローブのキャップ3個⼊り
T456C23956
フォイルセット：スケール1、0～1500µm（0～60mil） T99022255-1
フォイルセット：スケール1、0～1500µm（0～60mil）- T99022255-1C
証明書付き
フォイルセット：スケール2、0～5mm（0～200mil）
T99022255-2
フォイルセット：スケール2、0～5mm（0～200mil）- T99022255-2C
証明書付き
磁性⾦属⽤ゼロ点調整板、1%T9994910⾮磁性⾦属⽤ゼロ点調整板、1%T99949115 注意事項
スキャンモードと⾃動測定モードを使⽤できるのは、Elcometer 456
モデルTセパレート型膜厚計のみです。これらのモードでスキャンプ
ローブを使⽤する場合は、プローブの先端を傷つけないように必ずキャップを装着してください。キャップを装着せずにこれらのモード
でプローブを使⽤し、破損した場合は、保証の対象外となります。
スキャンプローブをElcometer 456セパレート型膜厚計に接続し、標準モードで使⽤する場合は、キャップを装着してもしなくてもかまいません。
スキャンプローブにキャップを装着する測定⽅法は、100µm（4mil）
より薄い膜厚の測定には適していません。

hキャップの厚さが200µm（8mil）より⼩さい場合.

jp-5

www.elcometer.com

R

www.elcometer.com

jp-6

R

TMA-0568-12第04版、本⽂と表紙番号23959