Precision Hydrogen Trace
Hydrogen Generator for GC
Part Number : See Reverse

Service Kit : See Reverse

Your local gas generation partner

Description
The Precision Hydrogen generators are designed to provide thehydrogen gas required for carrier gas and detectors requiring
Hydrogen fuel gas, such as FID and FPD. One generator iscapable of supplying multiple GCs, and can be suited to individual
customers’ needs, depending on the flow requirement.
These generators utilize a Proton Exchange Membrane (PEM) togenerate Hydrogen gas from deionized water, with a pressure
swing adsorption (PSA) dryer used to remove residual moisturefrom the gas.
Precision Hydrogen gas generators come with various safety
features as standard, giving you complete peace of mind in thelaboratory and a far safer, dependable and more convenient
alternative to cylinder gas.

Applications)

Carrier Gas• GC-MS• GC)

Detector Gas• Fuel Gas

Key Features

540 mm380 mm406 mm)

Suitable for flame gas and carrier gas at trace detection
limits)99.99999% Purity)

Internal leak detection with automatic shutdown features)

Proven PEM technology to generate hydrogen safely andreliably)

Regenerative PSA dryers to ensure highest level of purity)

Automatic loading pump as standard)

Maintenance limited to replacing de-ionizer cartridge)

Compact, space-saving modular design)

Creates hydrogen on demand, minimal storage of hydrogenin the system)

Combine multiple units for higher flow requirements)

GC in-oven hydrogen leak detector available as an optionalextra)

Peak offers a 3 year cell warranty with this generator asstandard.

Precision Hydrogen Trace shown in a typical Precision stack.

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Hydrogen Trace Data Sheet RSID 383 Rev 17 EN —08/07/2024

Technical Specifications

Hydrogen Trace, 250cc

Hydrogen Trace, 500cc

Hydrogen Trace , 1200cc

250 cc/min

500 cc/min

1200 cc/min

Max Flow Rate*
Max Pressure

100 psi / 6.9 bar

Purity99.99999%**

Dew Point-75 °C / -103 °F

Gas Outlets

1 x 1/8” Swagelok compression fitting

Water Purity Requirement<1.0μ Siemens/cm OR >1 Mohm-cm

Water Consumption

0.17 - 0.46 L/day

Operating Temperature

0.4 - 1.2 L/day

up to 2.1 L/day

10°C - 35°C / 50°F - 95°F

Electrical Requirements

110 / 230 V 50 / 60 HZ max 6 A

Power Consumption

477 Watts

110 / 230 V 50 / 60 HZ 10 A max

787 Watts

Heat Output

787 Watts

up to 1000 BTU/ hr

Generator Dimensions (WxDxH)

380 x 540 x 406mm / 14.9 x 21.2 x 15.9”

Generator Weight

29 Kg (64 lbs)

Noise Level

31.5 Kg (69.5 lbs)
Silent in operation* All flow rates are stated in sccm at 273.15K and 1.01bar** Based on O2 content independently verified by National Physical Laboratory, UK

Ordering Information
Part Number

64-025064-0500

3301768 (110v/230v)

Annual Service
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan

Accessories

Water Bottle 4L

Water Bottle 8L

Hydrogen Leak Detector (GC in-oven)Part Number

10-9016

10-901710-9010Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Hydrogen Trace Data Sheet RSID 383 Rev 17 EN —08/07/2024