3500/94M VGA Display Monitor
Datasheet

Bently Nevada Machinery Condition Monitoring

122M3217 Rev. L

Description
The 3500/94M VGA module displays all 3500 machinery
protection system information, including:
l
l
l
l
l

System event list
Alarm event list
All module and channel data
Alarm and OK status
Nine custom display options

Use the 3500 Rack Configuration Software to configure up tonine custom screens. By default, one bar graph screen shows
all of the 1X-filtered vibration measurements, while anothershows the proximity transducer gap values.
You can organize custom screens into machine train groups.
You can also organize system data into any pages on thecustom screen.

System Components
The 3500/94M is comprised of:lll

The 3500/94M VGA module
An I/O module
A 10" or 15" VGA touch screen display

You can mount the VGA display monitor using standard
cables up to 6 m (20 ft) from a 3500 rack, or using a
supported Keyboard Video Monitor (KVM) Extender, up to 10
km (6.2 miles) from the 3500 rack.

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Touch Screen Displays
Use either the 10" and 15" Advantech VGA touch screen displays. These compact, durable displays aredesigned for excellent viewing quality in industrial applications.
The 10" Advantech display is certified for hazardous environments. Each display requires a separate
power supply. The 3500 Rack Configuration Software supports both display drivers.

Reduce the Number of Displays
Using a standard cable and the optional Display Router Box, you can view data from up to four 3500
racks on one display.
The alarm and OK status is continually displayed in the upper right corner of the VGA display. A rack
select button changes color to indicate if any rack connected to the Display Router Box is Not OK or in
Alarm status.
Due to limitations imposed by the VGA standard, the Display Router Box must be locatedwithin 6 m (20 ft) of each connected 3500 rack. Longer cables are not recommended
because they may degrade image quality. Noisy electronic environments also degrade
image quality.

Access Data Remotely
If you need to display 3500 data at a greater distance, you have two options.
Distancefrom Display
Router Box

Options

Notes

Minor loss in image quality.
from 6.4–305 Wired CAT5 UTPm (21–1,000
Keyboard VideoThe receiver is certified for Class I, Division 2 hazardous areaft)
Monitor (KVM) Extender installations in North America.
When using the wired CAT5 UTP KVM Extender, you can tune
image quality using a PS/2 keyboard (not supplied) as
described in the 3500/94M VGA Display Module Operationand Maintenance Manual (document 122M3218).
No loss in image quality.
> 10 m –10 km Single-mode, fiber(33 ft–optic (SM FO) Keyboard Not certified for hazardous locations. Must be installed in a
6.2 mi)Video Monitor (KVM)
safe area or in an approved enclosure.
ExtenderOnly the wired KVM receiver is certified for Class I, Division 2 hazardous area installations inNorth America. The transmitter is not and must be installed in a safe area or enclosed in an
approved enclosure.

2/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

The SM FO KVM receiver and transmitter are not certified for hazardous area installations. Boththe receiver and the transmitter must be installed in a safe area or enclosed in an approved
enclosure.

3/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Specifications
3500/94M VGA Display Unit
InputsPower
4.6 watts max.
consumption
High-speed
network

Monitor status, data andvalues

RS232 port
(I/0)

Touch controller data

Outputs
OK LED

Indicates proper operation of
the module

TX/RX LED

Indicates communications
between other systemmodules

VGA port(I/O)

Analog red, green, blue andsync signals to the display

Front Port

24 Vdc Power out, VGA signal
out, Touch commands in.

Environmental Limits
Operating-20°C to 65°C
Temperature (-22°F to 149°F)
Storage-40°C to 85°C
Temperature (-40°F to 185°F)
Humidity

95%, non-condensing

4/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

WARNING

Compliance andCertifications

EXPLOSION HAZARD
Refer to User Manual
BEFORE making
connections.

This device complies with part 15 of the FCC Rules.
Operation is subject to the following two conditions:

DO NOT DISCONNECT
EQUIPMENT UNLESS POWER
HAS BEEN SWITCHED OFF OR
THE AREA IS KNOWN TO BE
NON-HAZARDOUS.

1. This device may not cause harmfulinterference.
2. This device must accept any interference
received, including interference that may
cause undesired operation.

EMC

AVERTISSEMENT - RISQUE D'EXPLOSION - AVANT DE
DECONNECTER L'EQUIPMENT, COUPER LE COURANT OUS'ASSURER QUE L'EMPLACEMENT EST DESIGNE NONDANGEREUX.

Standards
EN 61000-6-2
EN 61000-6-4
Directive
2014/30/EU

Electrical
SafetyRoHS

Standards
CSA C22.2 61010-1
UL EN 61010-1
Directive
2014/35/EU
Directive
2011/65/EU

WARNINGHAZARDOUS VOLTAGE
RISK OF ELECTRIC SHOCK,BURN OR DEATH.
Do not remove cover. No user
servicable parts inside.

5/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Advantech 10" Display (AN-D3510)

Hazardous Area Approvals

North America

For the detailed listing of country and productspecific approvals, refer to the Approvals Quick
Reference Guide (108M1756) available from
Bently.com.

CanadaEx ic nA IIC T4;
Certified to CSA C22.2 60079-0,11,-15

3500/94M VGA Display Module

Class I, Division 2, Groups A, B, C, D
T4;
Certified to CSA C22.2 213

Approval
Option (01)The following approvals apply to the3500/94M monitor, its I/O module,
and the 3500/94M Display Router Box.
North America

Class I, Zone 2, AEx ic nA IIC T4;
Conforms to: UL60079-0, -11, -15

CanadaEx nC [L] IIC T4 Gc:
Certified to CSA60079-0, -11, -15

ATEX/IECEx

Class I, Division 2, Groups A, B, C
and D:
Certified to CSA213

Approval
Option(02)

USA
Approval
Class I, Zone 2: AEx nC IIC T4 Gc:
Option (01) Conforms to: UL60079-0, -11, -15
Class I, Division 2, Groups A, B, C
and D:
Conforms to ISA 12.12.01

II 3G Ex ic nA IIC T4 Gc;-20°C ≤ Ta ≤ +65°C
Ex ic nA IIC T4 Gc; IP54;-20°C ≤ Ta ≤ +65°C

The Advantech 15" Display (FPM-8151)
is not certified to meet ATEX/IECEx
standards.

T4 @ -20 °C ≤ Ta ≤ +65 °C(-4 °F ≤ Ta ≤ +149 °F)
Per control drawing 149243
ATEX/IECEx

Approval
Option(02)

USAClass I, Division 2, Group A, B, C, D
T4;
Conforms to: ISA 12.12.01

II 3 GEx nA nC ic IIC T4 Gc
T4 @ -20 °C ≤ Ta ≤ +65 °C(-4 °F ≤ Ta ≤ +149 °F)
Per control drawing 149243

6/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Rack space

1 full-height front slot

I/O Module

Physical Characteristics

Rack space

1 full-height rear slot

3500/94M VGA Display Module

Keyboard Video Monitor (KVM) Extender

Dimensions(height x
width xdepth)

Mounting

Shelf top

Power input

12 Vdc and 5 Vdc

Agencies

UL certification for North
America only

Weight241.3 mm x 24.4 mm x 241.8 mm(9.5 in x 0.96 in x 9.52 in)
0.73 kg (1.6 lb.)

Temperature 0ºC to 45ºC
range
(32ºF to 113ºF)

I/O Module
Dimensions(height x
width xdepth)
Weight241.3 mm x 24.4 mm x 99.1 mm

Humidity(9.50 in x 0.96 in x 3.90 in)

The KVM Extender Receiver isapproved for Class I, Division 2
hazardous areas in North
America.

0.34 kg (0.75 lb.)

10" Advantech Display (ANA-D3510-100-A1)
Dimensions(height x
width xdepth)

249 mm x 389 mm x 48 mm

267 mm (10.5 in)

Weight4.0 kg (8.78 lb)

15" Advantech Display (FPM-8151)
Dimensions(width x
height x
depth)Class I,
Division 2

The KVM Extender Transmitter is
not certified for hazardous
usage and must be installed ina safe area or enclosed in an
approved enclosure (such as apurged housing or cabinet).

NEMA 4X

Not approved

9.8 in x 15.3 in x 1.9 inTouchscreen(diagonal)

To install in a hazardous area, see the3500/94M VGA Display Module
Operation and Maintenance Manual(document 122M3218).

422 mm x 338 mm x 68 mm
16.61 in x 13.31 in x 2.68 in

Display
screen(diagonal)

381 mm (15 in)

Weight8.5 kg (18.75 lb)

80%, non-condensing

Rack Space Requirements
Monitor Module

7/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

3500/94 Display Router Box
InputsPower +24
Vdc ± 6%

4.7 watts max

Input range

22.5 to 25.5 Vdc input

VGA inputs(4)Video image

RS232 inputs Touch screen control data
(4)
Outputs
OK status
LED

Indicates proper operation

VGA output(1)

Video image

RS232
output (1)Touch screen control data

8/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Dimensions(W x H x D)

387.4 mm (15.25 in) x 248.9 mm(9.8 in) x 49.0 mm (1.93
in)Panel Cutout: 342.9 mm(13.5 in) x 228.6 mm (9.0 in)

WeightShipping

4.0 kg (8.78 lb)

Warranty

One year

Display Monitor Specifications
Advantech 10” Display(ANA-D3510-100-A1)
Screen Size

10” diagonalType

LCD with LED back light

Industrial
hardened

Yes

Mounting
StylesFace Mount
19” Rack Mount
Panel Mount
Independent Mount

Power

24 Vdc power supply included.
(Permissible voltage range 10
Vdc to 28 Vdc)

Recommended for industrial use.
Certified for hazardous locations.
The 10" housing is designed for IP54
ingress protection against dust andwater spray.

Inrush Current: 3.5 A @ 24 V (<500 µs)Operating Current: < 500 mA
Agencies

CE, UL, CUL

IP54Designed for.

Class I, Div 2

Yes

ATEX/IECEx

Yes

Temperature-20°C to 65°C (-4°F to 149°F)
Operating Temperature.

The minimum temperature rating is-20ºC (-4°F) for the screen, front
bezel, and rear electronics. Whenthe screen or front bezel is exposed
to temperatures below -20ºC (-4°F),
a heated enclosure is required.
Humidity

10% to 85% RH noncondensing

TouchInterface

GFG Resistive Touch Screen

Manufacturer Advantech

9/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Advantech 15” Display(FPM-8151H)
Screen Size

15” diagonalType

LCD with LED backlight

Industrial
hardened

Yes

Mounting
Styles19” Rack Mount

Power

24 Vdc power supply included.
(Permissible voltage range 12
Vdc to 24 Vdc)

Warranty

Recommended for industrial use.
Not certified for hazardous
locations.

Panel Mount

Inrush Current: 2.62 A @ 24 V(< 500 µs)
Operating Current: < 500 mA
IP65

Yes

Agencies

CE, ULClass 1, Div 2

Yes

Temperature

0 to 60°C ( 32°F to 140°F)
Operating Temperature.

The minimum temperature rating is
0ºC ( 32°F) for the screen, front bezel,
and rear electronics. When thescreen or front bezel is exposed totemperatures below 0ºC ( 32°F) , a
heated enclosure is required.
Humidity

95% RH

Touch
Interface

Resistive

Manufacturer Advantech
Dimensions(W x H x D)
inches422 mm (16.61 in) x 338 mm(13.31 in) x 68.0 mm (2.68 in)

WeightShipping

8.5 kg (18.74 lb)

Two years

Panel Cutout: 396 mm (15.59
in) X 296 mm (11.65 in)

10/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

3. In the third and fourth racks don't installanother display, but install router box
cables for each additional rack. P/N
3500/94M-00-12-XX accommodates
these requirements.

Ordering Considerations
Firmware, Software, and Hardware
Requirements

To order a Display Router Box, use 3500/94M-AA
option 07.

The 3500/94M requires the 3500/22 Transient
Data Interface (TDI) module. It also requires
these or newer versions of the following
firmware and software:

Supported Display Modules
The 3500/94M supports these displays:l3500/94M Firmware

4.06, Rev. C

3500/22 Firmware (TDI)

Rev 1.71l

3500/01 Software (Rack
Config)

Rev 5.6l

3500/02 Software (OpDisplay)

Rev 2.21

3500/03 Software (Data
Acquisition)

Rev 1.22

The 3500/94M requires the following hardware:
3500/22 TDI - Transient Data Interface

Ordering a Display Router Box
Each 3500 rack requires both a
3500/94M VGA module andassociated I/O card. Because of thisrequirement, every AA ordering
option is structured to automatically
include these components.
To order multiple racks with one display unit:
1. In the first rack install a router box withrouter cable (P/N 3500/94M-07-12-XX).
This option also includes a power supply
cable for the router box.
2. In the second rack install a display unit
with router cable (P/N 3500/94M-(02 to06)-12-XX). These options also include a
power supply and power supply cable forthe display unit.

11/23

Advantech 10” touch screen display(P/N 120M8155-01)
Advantech 15” touch screen display(P/N 102M8950)
Parker touch screen displays (P/N148653-01 - no longer available).

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. Lm)Ordering Information
For the detailed listing of country andproduct specific approvals, refer tothe Approvals Quick Reference Guide(108M1756) available from Bently.com.

System Display

20

SM FO KVM SNGL extender andcable
2,000' (609.6 m)

21

SM FO KVM SNGL extender andcable
4,000' (1219.2 m)

22

SM FO KVM SNGL extender andcable
6,000' (1828.8 m)

23

SM FO KVM SNGL extender andcable
8,000' (2438.4 m)

24

SM FO KVM SNGL extender withpatch cables

25

SM FO KVM DUAL extender andcable
2,000' (609.6 m)

26

SM FO KVM DUAL extender andcable
4,000' (1219.2 m)

27

SM FO KVM DUAL extender andcable
6,000' (1828.8 m)

3500/94M-AA-BB-CC
A: Display Size and Mounting
00

None

01

10" display, face mount(only available for rackmounted 3500 monitor)

02

10" display, 19" rack mount03

10" display, panel mount(installation hardware
included)04

10" display, independent
mount

05

15" display, 19" rack mount06

15" display, panel mount(installation hardware
included)28

SM FO KVM DUAL extender andcable
8,000' (2438.4 m)

07

Router box, no display

29

SM FO KVM DUAL extender withpatch cables

B: KVM Extender and Display Router Box
Cables
00

None

01

Wired CAT5 UTP KVM extender andcable 250 ft (76.2 m)

02

Wired CAT5 UTP KVM extender andcable 500 ft (152.4 m)

03

Wired CAT5 UTP KVM extender andcable 1,000 ft (304.8 m)

12Display Router Box cable 20 ft (6C: Agency Approvals

12/23

00

None

01

CSA/NRTL/C (Class I, Division 2)

02

Multi (CSA/ATEX/IECEx) (onlyavailable for the 10" display)3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

the 10" Advantech display. Notethat cable assembly 119M650303 is also required from the3500/94M IO Module (19021402) to an industrial 24 VDC
power supply.

Accessories and Spares

119M6509-03

Power supply cable for 10"
Advantech Display thatconnects the rear of a rackmounted 3500/94M IO module(190214-02) to an industrial 24
VDC power supply. 10 ft (3.05m). Note that cable assembly
107M3696-01 is also required
from the front of the 3500/94M
Module to the display.

02200794

24V DIN mount power supply
with no cable

3500/94M Module Accessories andSpares122M3217

3500/94M VGA Display Monitor
Product Datasheet

122M3218

3500/94M VGA Display Module
Operation and Maintenance
Manual184826-01

3500/94M Main module

190214-01

3500/94M Normal I/O module

190214-02

3500/94M Power-only I/O module
for front-mount display

Other Mounting Accessories
122M3369-01

94M rack mount for Advantech
10” display

3500/94M Advantech 10" Display
Accessories

122M3370-01

94M panel mount for
Advantech 10” display

For display monitor specifications, See Display
Monitor Specifications on page 9.

122M3371-01

94M independent mount for
Advantech 10” display

119M6509-01

Power supply cable only for 10"
Advantech Display thatconnects from the display toan industrial 24 VDC power
supply. 10 ft (3.05 m).

120M8155-01

ANA-D3510-100-A1, Advantech
10” display. Does not include
mounting screws.

Front Mounting Accessories and Spares
122M3368-01

119M6509-02

94M face mount for Advantech
10” display
Mounting standoff for facemount display with threaded
mounting holes for 19" EIA rack.

122M4631-AA

161087-AA00

AA: mounting hole thread size.
Choose one:01 – M6 x 1
02 – 10-32
03 – M5 x 0.8

107M3696-01

148638-AA

3 ft (.91 m) v Video, serial, andpower cable from front of rackmounted 3500/94M Module to

VGA male-male cable fordisplay to 3500/94M I/O
AA: cable length. Choose one:
10 - 10 ft (3.05 m)
20 - 20 ft (6.1 m)
DB-9 male-female cable fordisplay to 3500/94M I/O or
3500/94 Display Router Box.
AA: cable length. Choose one:
01 - 10 ft (3.05 m)

13/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

149356-01

02 - 25 ft (7.62 m)

Display Router Box

Display Router Box Accessories and Spares

3500/94M Advantech 15" Display
Accessories

Split digital cable for display
router box for router to
3500/94M I/O

For display monitor specifications, See Display
Monitor Specifications on page 9.

AA: cable length. Choose one:

102M8950

149710-AABB

FPM-8151, Class I, Division 2, 15"
Advantech display. Includes
screws and hardware for panel
mounting.

BB: connectors. Choose one:
00 Assembled I/O connector
01 Unassembled I/O connector

Display Accessories and Spares

104M3812-AA

19” EIA mounting plate for theFPM-8151H 15" display (9U rack
mount)

161087-AA00

AA: panel color. Choose one:
01 RAL 7032 (Pebble Gray)
02 RAL 7035 (Light Gray)

161087-AA00

148638-AA

10 – 10 ft (3.05 m)
20 – 20 ft (6.1 m)

VGA male-male cable fordisplay to 3500/94M I/O

148638-AA

AA: cable length. Choose one:
10 - 10 ft (3.05 m)
20 - 20 ft (6.1 m)

VGA male-male cable fordisplay to 3500/94M I/O
AA: cable length. Choose one:
10 - 10 ft (3.05 m)
20 - 20 ft (6.1 m)
DB-9 male-female cable fordisplay to 3500/94M I/O or
3500/94 Display Router Box toAdvantech display.
AA: cable length. Choose one:
01 - 10 ft (3.05 m)
02 - 25 ft (7.62 m)

DB-9 male-female cable fordisplay to 3500/94M I/O or
3500/94 Display Router Box.

119M6509-04

Industrial DC power supply
cable to 3500/94 router box 10 ft(3.05 m).

AA: cable length. Choose one:
01 - 10 ft (3.05 m)
02 - 25 ft (7.62 m)

119M6509-02

Power supply cable for 15"
Advantech Display thatconnects from the display toan industrial 24 VDC power
supply. 10 ft (3.05 m).

02200794

24V DIN mount power supply
with no cable

Wired Category 5, Unshielded Twisted
Pair (CAT5 UTP) KVM Extender
Only the KVM receiver is certified for Class I,
Division 2 hazardous area installations inNorth America. The transmitter is not andmust be installed in a safe area or enclosed in
an approved enclosure.

123M3410-01

3500/94M Display Router Box

14/23

Pepperl + Fuchs KVM kit:includes transmitter, receiver,and power supply for thereceiver.

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

supply SM FO KVM transmitter

KVM CAT5 UTP Extender Accessories andSpares128M4542

CAT5 UTP cables
148350AAAA

138M0323

161087-AA00

148638-AA

Rose Electronics dual 5V power
supply SM FO KVM transmitter

AAAA: cable length. Chooseone:

KVM SM FO Extender Accessories and Spares0250 – 250 ft (76.2 m)
0500 – 500 ft (152.4 m)
1000 – 1000 ft (304.8 m)

128M4543

Rose Electronics 19” EIA rack
mount kit for SM FO KVM receiveror transmitter

VGA MF cable for KVM
transmitter to 3500/94M I/O. 15
ft (4.6 m)

128M4545

DVI-I to VGA cable for SM FO KVM
transmitter to 3500/94M I/O or SM
FO KVM receiver to display. 10 ft(3.05 m)VGA MM cable for KVM receiverto display

128M4552

SM FO patch cable 20 ft (6.1 m)

128M4548

SM FO cable 2000 ft (609.6 m)

128M4549

SM FO cable 4000 ft (1219.2 m)

128M4550

SM FO cable 6000 ft (1828.8 m)

128M4551

SM FO cable 8000 ft (2438.4 m)

AA: cable length. Choose one:10 – 10 ft (3.05 m)
20 – 20 ft (6.1 m)
DB-9 MF straight through serial
cable for KVM extendertransmitter to 3500/94M I/O or
KVM extender receiver todisplay
AA: cable length. Choose one:01 – 10 ft (3.05 m)
02 – 25 ft (7.62 m)

Single-Mode Fiber Optic (SM FO)
KVM Extender
The SM FO KVM receiver andtransmitter are not certified forhazardous area installations. Boththe receiver and the transmittermust be installed in a safe area or
enclosed in an approved enclosure.

128M4539

Rose Electronics single 5V power
supply SM FO KVM receiver

128M4540

Rose Electronics dual 5V power
supply SM FO KVM receiver

128M4541

Rose Electronics single 5V power

15/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Graphs and Figures
The examples in this section show included components and installation arrangements for various
ordering options.
Ordering options are structured as 3500/94M-AA-BB-CC, where:lll

AA = Display size and mounting
BB = KVM Extender, router box and cables
CC = Agency approvals

In addition to these options, all purchasing options automatically include a 3500/94M VGA display
module and appropriate I/O module. Separate components are available individually. See Ordering
Information on page 12.

Figure 1: Single Rack with One Face Mount Display

Ordering Option
3500/94M-01-00-00

Description
10" display with hinged face mount for EIA 19" rack

16/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Figure 2: Single Rack with One Display
Ordering Option

Description

3500/94M-02-00-(00/01/02)

10" display for mounting in 19" EIA rack

3500/94M-03-00 (00/01/02)

10" display, panel mount (installation hardware included)

3500/94M-04-00-(00/01/02)

10" display with independent mounting hardware

3500/94M-05-00-(00/01)

15" display for mounting in EIA 19" Rack

3500/94M-06-00-(00/01)

15" display, panel mount (installation hardware included)

17/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Figure 3: Single Rack with Wired (UTP CAT5) Keyboard Video Monitor (KVM) Extender
Ordering Option

Description

3500/94M-02-(01/02/03)-(00,01†)

10" display for mounting in 19" EIA rack, and KVM extender with either
250, 500, or 1000 ft cable

3500/94M-03-(01/02/03)-(00,01†)

10" display, panel mount (installation hardware included) and KVM
extender with either 250, 500, or 1000 ft cable

3500/94M-04-(01/02/03)-(00,01†)

10" display with independent mounting hardware and KVM extenderwith either 250, 500, or 1000 ft cable

3500/94M-05-(01/02/03)-(00,01†)

15" display for mounting in EIA 19" Rack and KVM extender with either
250, 500, or 1000 ft cable

3500/94M-06-(01/02/03)-(00,01†):

15" display, panel mount (installation hardware included) and KVM
extender with either 250, 500, or 1000 ft cable

18/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L† Engineering approval is required for Class I Division 2 UTP CAT5 KVM Extender Installations inNorth America. The KVM Extender transmitter must be installed in a safe area. ATEX/IECEx
approval is not available for the UTP CAT5 KVM Extender.

19/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Figure 4: Multiple Racks with Router Box to One Display

First Rack Option - router box
Ordering Option
3500/94M-07-12-(00/01/02)

Description
Router box and rack-to-router cables

Second Rack - display options
Ordering Option

Description

3500/94M-02-12-(00/01/02)

10" display for mounting in 19" EIA rack, and required cables

3500/94M-03-12-(00/01/02)

10" display, panel mount (installation hardware and required cables
included)

3500/94M-04-12-(00/01/02)

10" display with independent mount housing, and required cables

3500/94M-05-12-(00/01)

15" display for mounting in EIA 19" Rack, and required cables

3500/94M-06-12-(00/01)

15" display, panel mount (installation hardware and required cables
included)

20/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Third and Fourth Rack - router cable for each additional rack
Ordering Option
3500/94M-00-12-(00/01/02)

Description
Rack-to-router cable

Figure 5: Single-mode Fiber Optic (SM FO) Keyboard Video Monitor (KVM) Extender to One Display

21/23

3500/94M VGA Display Monitor
Datasheet

Ordering Option

122M3217 Rev. L

Single-mode Fiber Optic KVM Extender including:

3500/94M-XX-20-00

Single Power supply and 2000’ SM FO cable.

3500/94M-XX-21-00

Single Power supply and 4000’ SM FO cable.

3500/94M-XX-22-00

Single Power supply and 6000’ SM FO cable.

3500/94M-XX-23-00

Single Power supply and 8000’ SM FO cable.

3500/94M-XX-24-00

Single Power supply and a pair of 20’ SM FO patch cables.

3500/94M-XX-25-00

Dual Power supplies and 2000’ SM FO cable

3500/94M-XX-26-00

Dual Power supplies and 4000’ SM FO cable

3500/94M-XX-27-00

Dual Power supplies and 6000’ SM FO cable

3500/94M-XX-28-00

Dual Power supplies and 8000’ SM FO cable

3500/94M-XX-29-00

Dual Power supplies and a pair of 20’ SM FO patch cables.

The SM FO KVM receiver and transmitter are not certified for hazardous area installations. Boththe receiver and the transmitter must be installed in a safe area or enclosed in an approved
enclosure.

22/23

3500/94M VGA Display Monitor
Datasheet

122M3217 Rev. L

Copyright 2022 Baker Hughes Company. All rights reserved.

Bently Nevada and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes business, in the UnitedStates and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company. All other product andcompany names are trademarks of their respective holders. Use of the trademarks does not imply any affiliation
with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

23/23