Medical X-Ray High-Voltage Generator

DMP-100R / DMP-100RF / Stationary System
DMP-200 Series / Stationary System with Capacitor
DMP-300 Series / Mobile System• 3-phase adaptive input voltage improves image quality by reducing ripple and noise• 2-speed motor-starter support satisfies system design requirements• Connects to AEC/DAP peripherals for optimal machine configuration• Digital control circuit enhances performance and accuracy

Hospital

ClinicDELTA JOINS RE100

100% Renewable Electricity and Carbon
Neutrality Targets for Its Global Operationsby 2030.

ABOUT DELTA
Delta was founded in 1971 and has been the global leader in switching power supply solutions since 2002 andDC brushless fans since 2006. Delta offers some of the most energy efficient power products in the industry,including switching power supplies with efficient over 90%, telecom power with up to 98%, and PV inverters withup to 98.8% efficient. We have also developed the world's first server power supply certified as 80 Plus Titanium
with over 96% efficient. We regularly invest around 8% of our annual sales revenues in R&D and have worldwide

20152018

2021

Commitments for"We Mean Business"

Commitments for
EV100

Commitments forRE100• Science-based emissions reduction

Scope:• Use 100% renewable electricity

targets (SBT)• Climate change information in main
reports (TCFD)• Responsible corporate engagement

R&D facilities in Taiwan,China, Europe, India, Japan, Singapore,Thailand, and the U.S.

Delta's major operation sites

in climate policy

in global operations by 2030

Commitment:

Carbon Neutrality• Expansion of EV charging facilities• Achieve carbon neutrality• Switch to using EVs for company
vehicles by 2030
• Incentives for employees andcustomers to use EVsby 2030Race to Zero
• Signed business ambition

BUSINESS CATEGORIES

to meet the 1.5ºC target

Power ElectronicsAutomation

Infrastructure• Components• Industrial Automation• ICT Infrastructure• Power and System• Building Automation• Energy Infrastructure• Fan & Thermal Management• Automotive Electronics& Industrial Solutions• Display Solutions

FOCUSED ON SEVEN UN SUSTAINABLE
DEVELOPMENT GOALSDMP-100R
Series Generator
DMP-100R

32 kW

40 kW

50 kW

56 kW

65 kW

71 kW

80 kW

82 kW

Input
Input
Voltage

Input
Current

230VAC ± 10%, 1Ф
230/400 – 277/480VAC±10%, 3Ф
260A ± 10%@ 230VAC

325A± 10%@ 230VAC

80A ± 10%@ 400VAC

100A ± 10%@ 400VAC

70A ± 10%
@ 480VAC

88A ± 10%
@ 480VAC

230/400 – 277/480VAC ±10%, 3Ф

115A ± 10% @400VAC

129A ± 10%@ 400VAC

150A ± 10%@ 400VAC

164A ± 10%@ 400VAC

185A ± 10%@ 400VAC

190A ± 10%@ 400VAC

100A ± 10%@ 480VAC

112A ± 10% @480VAC

120A ± 10%@ 480VAC

132A ± 10%@ 480VAC

150A ± 10%@ 480VAC

154A ± 10%@ 480VAC

10mA 800mA

10mA 1000mA

10mA 1000mA0.1mAs 1250mAs

0.1mAs 1250mAs

0.1mAs 1250mAs

710mA /
100kV

800mA /100kV

800mA /103kV

InputFrequency

50Hz / 60Hz

OutputkV Range

40kV - 150kV, 1kV / StepkV accuracy± (5% of preset value + 1kV)

Rise Time< 1.5ms (10% - 90%)mA Range

10mA 800mA

10mA 630mA

10mA 800mA± (5% of preset value + 1mA)

ms Range

1ms - 10000ms

msAccuracy± (10% of preset value + 1ms)

Digital control circuit enhances performance and accuracy

mAs Range•

450-kHz high frequency performance•2-speed motor-starter supportmAsAccuracy•Connects to AEC / DAP peripherals•

Optional Console Support•

1 phase / 3 phase input power available

Output

10mA 500mAmAAccuracy•RADIOGRAPHY

10mA 400mA

Max. Output
Power0.1mAs 500mAs

0.1mAs 630mAs

0.1mAs 1250mAs

0.1mAs 1250mAs

0.1mAs 1250mAs± (5% of preset value + 0.2mAs)
320mA /
100kV

400mA /
100kV

500mA /
100kV

560mA /100kV

800mA /80kV

Hardware

Input

ExternalOutput
Voltage

24VDC ± 8% / 2A, 24VAC ± 10% / 6A, 110VAC ± 10% / 0.8A or 220VAC ± 10% / 0.8A

MotorStarter

Dual Speed Auto Control

SupportPeripherals

AEC / DAP / DRDimension

654 mm* 352 mm* 491 mm (L*W*H)

Weight75 kg

ConsoleOptional, APR: 64 Preset for User Programmable

491 mmStandard
SafetyIEC 60601-1 2nd edition, IEC 60601-1 3.1rd edition and IEC 60601-1 3.2rd edition, IEC 60601-2-54

EMC

IEC 60601-1-2 4ᵗʰ edition compliance

Certificate

654 mm4| www.deltaww.com

CE MDD(0197) / CE EMC / CB

352 mm| 5

DMP-100RF
Series Generator

DMP-100RF32 kW

40 kW

50 kW

56 kW

65 kW

71 kW80 kW82 kW

185A ± 10%@ 400VAC

190A ± 10%150A ± 10%@ 480VAC

154A ± 10%

Input
Input Voltage

Input Current

230VAC ± 10%, 1Ф
230/400 – 277/480VAC ±10%, 3Ф
260A ± 10%@ 230VAC

325A± 10%80A ± 10%

100A ± 10%@ 400VAC@ 400VAC

70A ± 10%
@ 480VAC@ 480VAC@ 230VAC

88A ± 10%230/400 – 277/480VAC ±10%, 3Ф

115A ± 10%@ 400VAC

129A ± 10%

100A ± 10%@ 480VAC

112A ± 10%

Input Frequency@ 400VAC@ 480VAC

150A ± 10%@ 400VAC

164A ± 10%120A ± 10%@ 480VAC

132A ± 10%@ 400VAC@ 480VAC@ 400VAC@ 480VAC

50Hz / 60Hz

Radiography Mode Output
kV Range

40kV - 150kV, 1kV / StepkV accuracy± (5% of preset value + 1kV)

Rise Time< 1.5ms (10% - 90%)mA Range

10mA - 400mA

10mA - 500mA

10mA - 630mAmA Accuracy

10mA 1000mA

10mA 1000mA± (10% of preset value + 1ms)
0.1mAs -0.1mAs -0.1mAs -0.1mAs -0.1mAs -0.1mAs -0.1mAs -0.1mAs -500mAs

630mAs1250mAs

1250mAs

1250mAs

1250mAs

1250mAs1250mAs

710mA / 100kV

800mA / 100kV

800mA / 103kV

mAs Accuracy
Power

10mA - 800mA

1ms - 10000ms

ms Accuracy

Max. Output

10mA - 800mA± 5% of preset value + 1mA

ms Range

mAs Range

10mA - 800mA± (5% of preset value + 0.2mAs)
320mA / 100kV

400mA / 100kV

500mA / 100kV

560mA / 100kV

800mA / 80kV

Pulsed Fluoro Mode Output
Output kV Range•

Support single, pulse and continuous mode

•Digital control circuit enhances performance and accuracy•450-kHz high frequency performance•2-speed motor-starter support•Connects to AEC / DAP peripherals•1 phase / 3 phase input power available

FLUOROSCOPY

40kV - 125kV, 1kV / Step

Output kV
Accuracy±10% of preset value

Output mARange

5mA - 100mA

Output mA
Accuracy±20% of preset value

Pulse Frequency

30, 15, 7.5, 3, 1 @ 60Hz; 25, 12, 6, 3, 1 @50Hz

Pulse width

1ms - 40ms in 1ms step, up to 50% duty cycle

Max. Average
Power6, 7.5, 12, 15 fps, Max average Po: 1.2kW; 25, 30 fps Max average Po: 1kWContinuous Fluoro Mode Output
Output kV Range

40kV - 125kV, 1kV / Step

Output kVAccuracy± 5% of preset value + 1kV

Output mARange

0.5mA - 20mA

Output mAAccuracy± 20% of preset value

HardwareInput

491 mmOutputExternal Output
Voltage

24VDC ± 8% / 2A, 24VAC ± 10% / 6A, 110VAC ± 10% / 0.8A or 220VAC ± 10% / 0.8A

Motor Starter

Dual Speed Auto Control

Support
Peripherals

AEC / DAP / DR

Dimension

654 mm* 352 mm* 491 mm (L*W*H)

Weight75 kgStandard

654 mm

6| www.deltaww.com

352 mmSafety

IEC 60601-1 2nd edition, IEC 60601-1 3.1rd edition and IEC 60601-1 3.2rd edition, IEC 60601-2-54

EMC

IEC 60601-1-2 4ᵗʰ edition compliance| 7

DMP-200
Series Generator
DMP-200

32 kW

40 kW

Input
Input Voltage

100 VAC - 240 VAC 1Ф

Input Current Range

15 A - 10 A

Input Frequency

50 Hz / 60 Hz

OutputkV Range

40 kV – 150 kV, 1 kV / Step

kV Accuracy± (5% of preset value + 1 kV)Rise Time< 1.5 ms (10% - 90%)mA Range

10 mA - 500 mA, R’20mA Accuracy+/- (5% of preset value + 1 mA)

ms Range

1 ms – 6300 ms, R’20

ms Accuracy
mAs
mAs Accuracy

42 kV / 100 mAs, 125 kV / 32 mAs, 32 kW / 100 ms
max, R’20

140 mAs (40 kW), 280 mAs max, R’20

± (5% of preset value + 0.2 mAs)

± (5% of preset value + 0.2 mAs)

Initial Charging Time

9 sec

35 secRecharge Time after Max.
mAs Output

5 sec

18 sec

320 mA / 100 kV

400 mA / 100 kV

24 VDC / 2 A, 24 AC / 6 A, 110 VAC / 2 A or 220
VAC / 1 A

24 VDC / 2 A, 24 VAC / 6 A, 110 VAC / 2 A or 220
VAC / 1 A

Dual Speed Auto Control

Dual Speed Auto Control

AEC / DAP, Optional

AEC / DAP, Optional

Max. Output Power•

Digital control circuit enhances performance and accuracy

Hardware•450-kHz high frequency performance

External Output Voltage•

Flat design for low-bed applications

Motor Starter•

Energy storage pack inside

Support Peripherals•

Universal 100–240 VAC input•

Connects to AEC / DAP peripherals•

External large capacitor bank (DMP-200 40kW)

CapacitanceDimension (W x H x D)
WeightConsoleDMP-200 32kW

± (10% of preset value + 1 ms)

60 mF

228 mF

630 mm x 400 mm x 370 mm

Generator: 630 mm x 370 mm x 400 mm
Capacitor Bank : 503 mm x 359 mm x 195 mm

60 kg

Generator: 60 kg
Capacitor Bank: 25.5 kg

Optional, APR: 192 Preset for User Programmable

DMP-200 40kW

StandardSafety

IEC 60601-1 2nd edition, IEC 60601-1 3.1rd edition and IEC 60601-1 3.2rd edition, IEC 60601-2-54

Generator

EMC

IEC 60601-1-2 4ᵗʰ edition compliance

Certificate

CBCapacitor Bank8| www.deltaww.com| 9DMP-300
Series Generator
DMP-300

32 kW40 kW50 kW

Input
40 - 60 VDC, for Generator
100 VAC – 240 VAC 1Ф, for Battery Charger

Input Voltage
Input Current Range

15 A - 10 A, for Battery Charger

Input Frequency

50 Hz / 60 Hz, for Battery Charger

OutputkV Range

40 - 150 kV, 1 kV / Step

kV Accuracy± (5% of preset value + 1 kV)Rise Time< 1.5 ms (10% - 90%)mA Range

10 mA - 630 mA, R’20

mA Accuracy+/- (5% of preset value + 1 mA)

ms Range

1 ms - 10000 ms, R’20

ms Accuracy± (10% of preset value + 1 ms)

mAs Range

0.1 mAs - 630 mAs @40kV, R’20mAs Accuracy± (5% of preset value + 0.2 mAs)

Max. Output Power

320 mA / 100 kV / 160 ms

500 mA / 100 kV / 100 ms
560 mA / 100 kV / 20 ms

400 mA / 100 kV / 125 ms

HardwareBattery Charger
Battery Pack
Energy Storage Pack•

50kW output reduces exposure time for higher-quality images•

High-capacity 3U-NCM Li-based battery extends operation time•

12/24/105 VDC auxiliary power simplifies system integration•Digital control circuit enhances performance and accuracy•

Built-in energy storage system extends the battery life

External Output Voltage
Motor Starter

52 VDC / 19A
NANA

3U-NCM Battery (48V, 50Ah)

126 mF
12 VDC / 14A, 24 VDC / 10A, 105 VDC / 5A
Dual Speed

Support Peripherals

AEC / DAP, Optional

Dimension (L x W x H)

High-Voltage Generator: 500 mm x 352 mm x 305 mm
Battery Pack: 460 mm x 430 mm x 120 mm
Energy Storage Pack: 500 mm x 352 mm x 150 mm

WeightHigh-Voltage Generator: 22.0 kg
Battery Pack: 28.5 kg
Energy Storage Pack: 18.5 kg

StandardSafetyElectromagnetic Compatibility

IEC 60601-1-2 4ᵗʰ edition compliance

150 mm

305 mmBattery Pack

460 mmHigh-Voltage Generator

IEC 60601-1 2nd edition, IEC 60601-1 3.1rd edition and IEC 60601-1 3.2rd edition, IEC 60601-2-54

500 mm

350 mm
470 mmEnergy Storage Pack10

| www.deltaww.com| 11

Delta Electronics, Inc.
16 Tungyuan Rd., Chungli Industrial Zone, Taoyuan
City 32063, Taiwan
TEL : +886 3 452 6107
E-mail : x.ray@deltaww.com

www.deltaww.com

2023 / 02© 2023 Delta Electronics All Right Reserved. All information and specifications are subjected to change without prior notice.