Genius XE Nitrogen

High performance nitrogen generator for LC-MS
Part Number : 99-9999

Service Kit : 99-9999

Your local gas generation partner

Description
Inspired by the success of our best-selling Genius line of nitrogengas generators for LC-MS, Genius XE Nitrogen is a cutting-edge
evolution combining advanced technology with refined and robust
engineering. With two models - XE 35 (up to 35 L/min) and XE
70 (up to 70 L/min) - Genius XE Nitrogen provides a premium
standalone nitrogen solution for high performance LC-MS andother mission-critical laboratory applications where performanceand reliability are paramount.
Featuring Multi-Stage Purification™ and next-generation
integrated compressors with Electronic Compressor
Optimization™ (ECO) technology, Genius XE delivers factory
certifiable purity up to 99.5% on-demand, 24/7 with a convenient
fixed annual service interval.

Applications
On-demand nitrogen source for:)

LC-MS including new high-sensitivity instruments)

ELSD)

Small sample evaporators)

Other laboratory applications requiring nitrogenKey Features)

Variable flow up to 70 LPM)

Variable pressure up to 116 psi)

Multi-Stage Purification™ producing analytical grade
nitrogen gas up to 99.5% purity)

2 year comprehensive manufacturer’s warranty*)

Touch-screen full colour user interface for ease ofoperation)

ECO (Electronic Compressor Optimisation™)
technology for low energy consumption andcompressor durability)

Next-generation high performance premiumcompressors, engineered exclusively for Genius XE)

Smaller & quieter than previous generation of Genius)

On-board service scheduling and system diagnostics)

Additional compressor capacity for lower flow
applications (XE 70 only)

710 mm / 28”

1000 mm / 39.4”

650 mm / 25.6”

570 mm / 22.5”

Copyright © Peak Scientific Instruments Ltd 2023 — Genius XE Data sheet Rev 8 RSID 2463 EN —04/09/2023

Technical Specifications

Genius XE 35

Genius XE 70

Max Flow Rateup to 35 L/min

up to 70 L/min

Max Pressure

116 psi (8 bar)

116 psi (8 bar)Gas Outlets

1 x ¼” BSPP

1 x ¼” BSPP

Purity**95-99.5%

95-99.5%

80% RH @ 31°C

80% RH @ 31°C

Max Altitude

3000m

3000m

Particles< 0.01 µm< 0.01 µm

Phthalates

Phthalate & BHT Free

Phthalate & BHT Free

None

None

< 1 ppm NMHC

< 1 ppm NMHC

Operating Temperature5°C (41°F) to 35°C (95°F)

5°C (41°F) to 35°C (95°F)

Electrical Requirements

120VAC, 60Hz, 12A or 230VAC, 50/60Hz, 8A

230VAC, 50/60Hz, 12A

960 VA (120V)/1265 VA (230V)

2530 VA

4118 BTU

8785 BTU

56 dB

59 dB

650 x 570 x 710 mm
25.6 x 22.5 x 28 ”

1000 x 570 x 710 mm
39.4 x 22.5 x 28 ”

92 kg / 202 lbs

147 kg / 325 lbs

3300252 (230VAC) /3300807 (120VAC)

3300253

Max Relative Humidity

Suspended Liquids
Non-Methane Hydrocarbon Content

Power Consumption
Heat Output
Noise Level†

Generator Dimensions (H x W x D)
Generator Weight

Ordering Information
Part Number
Annual Service
Complete Maintenance Planvisit: www.peakscientific.com/ordering** Factory certified purity with respect to O2 content. Actual purity varies in relation to flow (min. 95% at maximum flow output)†Noise level expressed as SPL (Sound Pressure Level) measured at 1m from source in a reverberant chamber

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 2 year warranty*. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.
*Year 2 of warranty subject to generator being serviced at end of year 1 by a Peak-approved agent in accordance with fixed annual maintenance schedule.
For full terms and conditions visit www.peakscientific.com/warranty-statement

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Peak Scientific’s Quality
Management System
conforms to:ISO 9001:2015

Product Certifications

For a full list of our worldwide
office locations, please visit:www.peakscientific.com/offices
Web: www.peakscientific.com
Email: discover@peakscientific.com

Copyright 2023 Peak Scientific - Rev - Genius XE Data sheet Rev 8 RSID 2463 EN - 04/09/23