Corona 1010
Nitrogen Generator
Part Number : 81-4000

Service Kit : 08-9013

Your local gas generation partner

Description
Peak’s Corona 1010 model has been developed to meetthe demands of labs requiring high purity nitrogen for theirapplications.
Suitable primarily for Charged Aerosol Detectors, the Corona
1010 generator delivers a high-purity, filtered nitrogen supply for
applications requiring flows of up to 5l/min at 80psi.
With a space-saving design, the Corona 1010 can fit seamlessly into
any laboratory. Engineered with an optional, dedicated compressor
module for customers with no suitable supply of in-housecompressed air, the Corona system provides a flexible solution forlabs with differing configurations.

Applications))

ThermoFisher ScientificCorona Veo))

ThermoFisher Scientific
Vanquish CAD

540 mmKey Features

417 mm
156 mm
406 mmCorona 1010 shown in a stack with Corona Air Compressor.

Copyright 2015 Peak Scientific - Corona 1010 Data Sheet Rev 7 RSID 1287 EN - 10/10/19))

Suitable for applications requiring high purity nitrogen))

Available with or without external compressor module))

Dedicated solution for Thermo Scientific’s Corona andVanquish instruments))

Tried and tested membrane technology delivers highpurity nitrogen))Gas is supplied on demand so generator works to yourschedule))

The most cost-effective method of gas supply

Technical Specifications

Corona 1010

Max Flow Rate

5 lpm

Max Pressure

80 psi / 5.5 barPurity<2% 02 @ 5 L/min

Gas Outlets

1 x 1/4 " BSPP

Min Air Inlet Pressure

7.6 bar / 110 psi

Min Air Inlet Flow

30 lpmMin Inlet Air Quality

ISO8573 - 1:2010 Class 1.4.1

Phthalates

None

Suspended Liquids

None

Start-Up Time For Purity

5 minutes

Operating Temperature

5°C - 35°C / 41°F - 95 °F

Electrical Requirements

100-230V 50/60 HZ 0.01 A

Power Consumption

10 Watts

Heat Output

Up to 140 BTU/Hr

Generator Dimensions (WxDxH)

156 x 417 x 540 mm /6.2 x 16.5 x 21.3’’

Generator Weight

12 Kg (26.5 lbs)

Ordering Information
Part Number

81-4000

Annual Service

08-9013

Standard Maintenance Plan

09 - 3110

Complete Maintenance Plan

09 - 3010

Corona Air Compressor

230V

110V

Part Number

85-0555

85-1555

Annual Service

08-8352

Generator Dimensions (WxDxH)
Generator Weight406 x 417 x 540 mm / 16 x 16.5 x 21.3’’
41.5 Kg (91.5 lbs)

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2015 Peak Scientific - Corona 1010 Data Sheet Rev 7 RSID 1287 EN - 10/10/19