3Thickness Gauge
High-precision measurement for papers,
films and parts thickness

Dial Indicator is used by being fitted to
jig etc., while thickness gauge is heldwith our hand. Holding work piece
between stylus and anvil, read the value
directly. Contact point moves to upward
when lifting lever is pressed down, andcontact point returns to “zero” when it isreleased. As operation is easy, it canmeasure for a short period comparedwith micrometer. There are 2 kinds ofDial 0.01mm , 0.001mm for both analog
and digital. The stroke depends on sizeof work piece and a model is available tomeasure maximum thickness up to
50mm. This can be used for various
thickness measurement such as paper,
hair, rubber plate metal tube small
molded components.

Pointer

Lifting Lever

Dial

Short短針
Hand

SpindleContact Point

Anvil
Flame

Measuring metal work piece.The photo shows
5.98mm.

Dial Thickness Gauge ------------------------ 66~68

Bluetooth Digital Thickness Gauge ------------- 72·73

Dial Pipe Gauge 0.01mm Model ------------- 69

Bluetooth Digital Constant Pressured Thickness Gauge -- 76

Dial Swift Gauge ----------------------------------- 68
Digital Pipe Gauge 0.01mm Model ----------- 69
Conventional Digital Thickness Gauge ------- 70
Standard

DigitalThicknessGauge------- 71

Constant Pressured Thickness Measuring Instrument --- 74·75

Thickness Gauge Special Product List ---- 77
Parts & Accessories ----------------------------------- 78

• As to shape of contact point and anvil, there are standard
type and other various kinds.

SM-112Standard type

Dimensions·Graduation 0.01mm·Measuring Range 10mm

104SM-112D(Coaxial Revolution Pointer)12 18.3

SM-112P(Direct Reading Graduation)

3

Dial Thickness Gauge• Suitable for measuring thickness and diameter of metal, lens,rubber, plastic, paper, felt, hair and pearl etc. In actual dimension.
• Ceramic contact point and anvil feature are superior foranti-abrasion and dust. In addition, there are steel FE type andAT type which rarely adheres with adhesion tape.

102

Thickness Gauge

Dial Thickness Gauge

26

All SM series equips ceramic contact
point and anvils.

64
unit : mm

Specifications
Model

Graduation(mm)

SM-1120.01

SM-112LW

0.01

SM-112LS

0.01

SM-112-3A

0.01

SM-112-80g

0.01

SM-112P

0.01

SM-112FE

0.01

SM-112AT

0.01

SM-112D

1.17mm reading example

SM-112 Series

0.01

Measuring
Range
(mm)

Indication Parallelism
Dial Reading
Error(μm)
(μm)

10

±15

10

±15

10
10
10
10
10
10
10

5

0-50-100

2.5 or less−

0-50-100

2.5 or less

±15

−

±15

5

±15
±15

150

φ3.2 Ball

φ3.2 Ball

150

φ10 Flat

φ5 Flat

2.5 or less

φ 5 Flat

φ10 Flat

φ10 Flat50-50-100

2.5 or less

φ10 Flat

φ10 Flat50-50-1000-0.5-1

8±15

φ10 Flat

Stop Point Measuring Forced 0.8±0.05

5±15

φ 10 Flat
φ3.2 Ball

2.5 or less

0-50-1005±150-50-100
0-50-100

Contact Point
Anvil Form Weight
Form
(mm)
(g)
(mm)

Measuring Force(N)

2.5 or less

φ10 Flat

0.8 or less

0-50-100

φ10 Flat

φ10 Flat

2.5 or less

φ10 Flat

φ10 Flat

φ10 Flat

Suggested
List Price
9,680 JPY

150

10,340 JPY

150

10,670 JPY

150
150

10,340 JPY
13,090 JPY

9,680 JPY

150

10,890 JPY

155

11,880 JPY

150

14,190 JPY

LS, LW, 3A For more information please refer to P77.

Dimensions
8 (15.8)

12

φ5

26

4
3

10

30
(6)

Anvil

Anvil

15

Contact
Point

20·Graduation 0.01mm·Measuring Range 20mm
·Ceramic Contact Point and Anvil(125.3)

Measurement of up to
20mm thick

φ55

Measuring Range

SM-528(70.5)
unit : mm

Specifications
Model
SM-528SM-528LS

SM-528LW

SM-528-3A

SM-528-80g
SM-528FE

SM-528 Series

Graduation Measuring
Range
(mm)
(mm)
0.01
20
0.01

20

0.01

20

0.01
0.01
0.01

20
20
20

±20

5
−

0-50-100

3.5 or lessContact Point Anvil Form Weight
Form
(g)
(mm)
(mm)
φ 10 Flat
φ 10 Flat
180

±20

5

0-50-100
0-50-100

3.5 or less

φ 3.2 Ball

φ 3.2 Ball

φ10 Flat

φ10 Flat

Indication Parallelism
Dial Reading
Error(μm)
(μm)
±20
±20
±20
±20

−
5
5

0-50-100

0-50-100
0-50-100

Measuring Force(N)

φ 3.2 Ball

3.5 or less
3.5 or less

Stop Point Measuring Forced 0.8±0.05
3.5 or less

φ 5 Flat

φ10 Flatφ 10 Flat

180

φ 5 Flat180

φ10 Flat

180
180
180

Suggested
List Price
13,530 JPY
14,190 JPY

14,190 JPY
14,630 JPY
17,710 JPY
14,740 JPY

LS, LW, 3A For more information please refer to P77.

66

4
28
15
3

·Graduation 0.01mm·Measuring Range 10mm
·Ceramic Contact Point and Anvil

30

φ10

26

Insertion Depth
120mm

108

66

SM -114120

157

Dial Thickness Gauge132

SM-114Thickness Gauge

Dimensions

unit : mm

Dimensions

SM-124

108
30

4

26

φ10

28
15
3

·Graduation 0.01mm·Measuring Range 20mm
·Ceramic Contact Point and Anvil

65

Medium size
Thickness Gauge132

SM -124120

157

SM-125

unit : mm

Large size Thickness
Gauge
Dimensions

SM-125

147

20

(32)

3

φ10

19

25

28
4

74.4
71.4
16

178

·Graduation 0.01mm·Measuring Range 20mm
·Ceramic Contact Point and Anvil
with Stand

6.5

This stand is standard accessories.

200

263.5
unit : mm

Specifications

SM-114 Series
Model

Graduation(mm)

SM-1140.01

SM-114LW

0.01

SM-114LS

0.01

SM-114P

0.01

Graduation(mm)

SM-1240.01

SM-124LW

0.01

SM-124LS

0.01

Graduation(mm)

SM-1250.01

SM-125LW

0.01

SM-125LS±15−0-50-100

2.5 or less

φ3.2 Ball

10

±15

5

0-0.5-1

2.5 or less

φ10 Flat

10

±15

−

0-50-100

2.5 or less

φ3.2 Ball

φ10 Flat

250

φ10 Flat

250

φ3.2 Ball

250

10,890 JPY
12,980 JPY
12,980 JPY
12,430 JPY

Measuring
Measuring Force Contact Point Form Anvil Form Weight
Indication Parallelism
Range
Dial Reading
Error(μm)
(N)
(mm)
(mm)
(g)
(μm)
(mm)
3.5 or less
φ10 Flat20
φ10 Flat±20250
50-50-100
20
20

±20
±20

−
−

0-50-100
0-50-100

3.5 or less
3.5 or less

φ3.2 Ball
φ3.2 Ball

φ10 Flat

φ3.2 Ball

250
250

Suggested
List Price
15,510 JPY
18,370 JPY
18,370 JPY

LS, LW, 3A For more information please refer to P77.

SM-125 Series
Model

10

Suggested
List Price

LS, LW, 3A For more information please refer to P77.

SM-124 Series
Model

Measuring
Measuring Force Contact Point Form Anvil Form Weight
Indication Parallelism
Range
Dial Reading
Error(μm)
(N)
(mm)
(mm)
(g)
(μm)
(mm)
2.5 or less
φ10 Flat10
φ10 Flat250±150-50-10050.01

Measuring Indication Parallelism
Measuring Force Contact Point Form Anvil Form Weight
Range
Dial Reading
Error(μm)
(N)
(mm)
(mm)
(g)
(μm)
(mm)
3.5 or less
φ10 Flat20
φ10 Flat±20625
0-50-100
5
20
20

±20
±20

−
−

0-50-100
0-50-100

3.5 or less
3.5 or less

φ3.2 Ball
φ3.2 Ball

φ10 Flat

φ3.2 Ball625
625

Suggested
List Price
38,720 JPY

40,700 JPY

40,700 JPY

LS, LW, 3A For more information please refer to P77.

67

Thickness Gauge

SM-1201L·Graduation 0.001mm·Measuring Range 1mm

Anvil·Graduation 0.001mm·Measuring Range 10mm
·Indication Range 1mm(Lifting Anvil)

Anvil Fixing
Screw·Contact Point,
Anvil = Solid Carbide

Symmetrical Dial

Carbidecontact point

SM-1202L·Graduation 0.001mm·Measuring Range 2mmContinuous Dial

·Contact Point,
Anvil = Ceramic·Contact Point,
Anvil = Ceramic*Setting up standard point with blockgauge is necessary to measure
thickness 1mm and over.

Dimensions

Model
SM-1201

SM-1201LS

SM-1201LW
SM-1201L
SM-1202L

34

30

φ10
30

Lifting of Anvil

70.5(72.5)

SM-1201L

unit : mm

SM-1201 Series

Graduation Measuring
Range
(mm)
(mm)
0.001
10
0.001

10

0.001

1(3)*

0.001
0.001

10

2(2)*

Indication
Error(μm)

Dial Reading

30-100-0

Measuring Contact Point Form
Anvil Form
Weight Suggested
Force
List Price
(mm)
(mm)
(g)
(N)
37,730 JPY
1.5 or less φ8.5 Flat (Carbide) φ8.5 Flat (Carbide) 440−

0-100-0

1.5 or less

±3
±3

−

±3

3

±3

0-100-00-100-200

3±50-100-200

φ8.5 Flat (Carbide)440

42,020 JPY

1.5 or less φ10 Flat (Ceramic) φ10 Flat (Ceramic)

420

27,830 JPY

1.5 or less

φ3 Ball (Carbide)φ3 Ball (Carbide)φ3 Ball (Carbide)

45
185

φ10
Lifting

74*Setting up standard point with block gauge is necessary to measure
thickness 30mm and over.

Specifications

SM-130LW

21

95
116

SM-130 Series

unit : mm

Measuring Contact Point Anvil Form Weight Suggested
Graduation Measuring Indication Parallelism
Dial Reading Force (N)(mm)
Form(mm)
List Price(mm) Range (mm) Error(μm) (μm)(g)
φ10 Flat50±25
5
620 38,390 JPY
0.01±0-50-100 2.2 or less φ 10 Flat
0.01
0.01

50
50

Dial Swift Gauge

±25
±25

−
−

±0-50-100 2.2 or less±0-50-100 2.2 or less

φ 3.2 Ball
φ 3.2 Ball

φ10 Flat

φ3.2 Ball

620
620

42,790 JPY
42,790 JPY

LS, LW, 3A For more information please refer to P77.

Dimensions
20

SM-130LS45

18·Upward Shockproof Contact Point, Anvil = Ceramic

SM-13030,250 JPY

Dimensions·Graduation 0.01mm·Measuring Range 50mm
·Indication Range 30mm
(Lifting Anvil)

Anvil Adjusting Screw

420

42,020 JPY

LS, LW, 3A For more information please refer to P77.

By the lifting of the anvil, canbe measured 0~50mm

Model

440

1.5 or less φ10 Flat (Ceramic) φ10 Flat (Ceramic)

SM-130

Anvil

unit : mm

Parallelism(μm)* ( ) is a free-stroke.

Contact
Point(72.5)

SM-1202L

unit : mm

120

Specifications(100.5)(99.4)(99.4)
34

9
18

SM-1201

φ10

6 3 6.5

134
25

30

11

φ8.5

Dimensions(100.5)6 3 6.5

Dimensions

20 of Anvil

Anvil Adjusting
Screw

58.5

Dial Thickness Gauge / Dial Swift Gauge

SM-1201

139
φ10

45

26

·Upward Shockproof·Contact Point, Anvil = Ceramic

31.5·Graduation 0.01mm·Measuring Range 20mm

68.5

SFM-627

159.5

Push top point down and nip work piece for measurement.

Specifications
Model
SFM-627

68

76

unit : mm

Graduation Measuring Indication Parallelism
Contact Point Anvil Form Weight Suggested
Dial Reading
(mm)
(mm)
Range (mm) Error(μm)(μm)
Form(mm)
List Price(g)
φ10 Flat
φ10 Flat±20
50-50-1000.01
20
240 23,760 JPY

Thickness Gauge

Dial Pipe Gauge

TPM-617·Graduation 0.01mm·Measuring Range 10mm

Dial Pipe Gauge / Digital Pipe Gauge

TPM-116*Suitable for measuring thickness of pipe andcurved plate etc. Radial thickness can bemeasured up to minimum diameter φ 3.5mm.

TPM- 618·Graduation 0.01mm·Measuring Range 10mm

·Graduation 0.01mm·Measuring Range 10mm

·Anvil replaceable type (Φ0.5,Φ1.0,Φ2.0mm)·Anvil replaceable type (option)·Upward Shockproof Anvil fixed type
Edge of point φ2.5mm Flat

Dimensions:TPM-617/618

59
unit : mm

The special order if thefollowing hole diameter
φ3.5mm.

(97.9)(41.5)(126.1)
22

35.4

Anvil

Ball Anvil φ3.5mm

Contact
Point

5

42.7
25.3

Contact
Point

(5)

103

(44.2)

Dimensions:TPM- 116

Anvil

14

(66)

Relation between Anvil diameterand Work inserting depth

Anvil dia.
φ0.5

Depth
2

Anvil dia.
φ5.0

φ2.0

3

φ10.0

φ1.0

unit : mm3

φ7.0

Graduation(mm)

TPM-1160.01

TPM-618

0.01

TPM-617

0.01

Measuring
Indication Parallelism
Range
Dial Reading
Error(μm)(μm)(mm)
0-50-100−10±150-50-100
±15
−
10
±15

10

−

0-50-100

MeasuringForce(N)
2.3 or lessContact Point
Form
(mm)
φ2.5 Flat

1.5 or less

φ1.6 Ball

1.5 or less

φ1.6 Ball

88unit : mm

Specifications
Model

Depth
8

Anvil Form
(mm)

Weight(g)

Suggested
List Price

φ3.5 Ball

145

10,340 JPY

φ5.0 (φ7.0,10.0 replaceable)*

195

30,250 JPY

φ0.5,1.0,2.0 replaceable

190

35,090 JPY*Anvils of φ7 and φ10.0 are optional.

Digital Pipe Gauge(97.9)

41.5
5

35.4

120

38.1

Dimensions:TPD - 617A/618A

Contact
Point

Anvil

14
(66)

TPD-617A·Resolution 0.01mm·Measuring Range 12mm

·Anvil replaceable type

·Anvil replaceable type (option)

TPD-617A
TPD-618A

Anvil dia.
φ0.5

Depth
2mm

Anvil dia.
φ5.0

Depth
8mm

φ2.0

3mm

φ10.0

8mm

φ1.0

3mm

φ7.0

8mm

unit : mm

Specifications
Model

Relation between Anvil diameterand Work inserting depth

TPD-618A·Resolution 0.01mm·Measuring Range 12mm

unit : mm

Resolution
(mm)
0.01
0.01

Measuring Range Indication Parallelism Measuring Contact Point Form
Error(μm)
(μm)
(mm)
(mm)
Force(N)
12
12

±20
±20

−
−

1.5
1.5

φ1.6 Ball

φ1.6 Ball

Anvil Form
(mm)

Weight(g)

φ5.0 (φ 7.0,10.0 replaceable)*

260

φ0.5,1.0,2.0 replaceable

255

Suggested
List Price

59,290 JPY
54,450 JPY*Anvils of φ7 and φ10.0 are optional.

69

• Digital display for error-free reading• Measurement force cannot be changed.Low measurement force is required, check a standard type.

Conventional Digital Thickness Gauge

SMD-540S2

φ107Insertion Depth 30mm Model

Standard Type

·Resolution 0.01mm·Measuring Range 12mm
·Ceramic Contact Point and Anvil

30

3

6.5

(66.9)(43.4)

Dimensions

30.5

Thickness Gauge

Conventional Digital Thickness Gauge

72.5
unit : mm

Specifications
Model
SMD-540S2

SMD-540S2-LS

SMD-540S2-LW
SMD-540S2-3A

Resolution Measuring Indication Parallelism(mm)Range(mm) Error(μm)
(μm)
0.01

12

±20

0.01

12

±20

0.01
0.01

12
12

5

±20

−

±20

5

−

Measuring Force(N)

Contact Point
Form(mm)

Anvil Form
(mm)

Weight(g)

2.0 or less

φ3.2 Ball

φ10 Flat

250

2.0 or less

φ5 Flat

2.0 or less
2.0 or less

φ10 Flat

φ3.2 Ball

φ10 Flat

φ3.2 Ball
φ5 Flat

Suggested
List Price250

26,950 JPY

250

31,900 JPY

250

30,030 JPY
33,440 JPY

LS, LW, 3A For more information, refer to P77.

15

110

φ10

120

3

18

25

75.4

151.8

Dimensions

Standard Typeunit : mm

SMD-550S 2

Insertion Depth 120mm Model

Resolution 0.01mm
Measuring Range 12mm
Ceramic Contact Point and Anvil

Specifications
Model
SMD-550S2

SMD-550S2-LS

SMD-550S2-LW
SMD-550S2-3A

Resolution Measuring Indication Parallelism
(mm)
(μm)
Range(mm) Error(μm)
0.01

12

±20

0.01

12

±20

0.01
0.01

12
12

5

±20

−

±20

5

−

Measuring Force(N)

Contact Point
Form(mm)

Anvil Form
(mm)

Weight(g)

2.0 or less

φ3.2 Ball

φ10 Flat400

2.0 or less

φ5 Flat

2.0 or less
2.0 or less

φ10 Flat

φ3.2 Ball

φ10 Flat

φ3.2 Ball
φ5 Flat

Suggested
List Price

400

27,940 JPY

400

32,890 JPY

400

31,020 JPY
34,320 JPY

LS, LW, 3A For more information, refer to P77.

70

Thickness Gauge

Standard Digital Thickness Gauge• 0.01mm and 0.001mm resolution are available.

Standard Digital Thickness Gauge

Anvilfixing screw
Anvil adjustingscrew

SMD-565A

SMD-540A

SMD-550A·Resolution 0.01mm·Measuring Range 12mm
·Ceramic Contact Point and Anvil·Resolution 0.01mm·Measuring Range 12mm
·Ceramic Contact Point and Anvil

Insertion Depth 30mm Model

By the lifting of the anvil,
can be measured 0-15mm·Resolution 0.001mm·Measuring Range 15mm
·Indication Range 12mm
(Lifting of anvil)·Ceramic Contact Point and Anvil

Dimensions:SMD-565A

Insertion Depth 120mm Model

Dimensions:SMD-550A

30(72.5)

7

unit : mm

30

72.5

3 12.5

30.5

18

3

25

41

17

110

144
72.9
φ10

3

4

101

64.4

(99)

φ10

27

4

63.4

38.1

38.1

Dimensions:SMD-540A

φ10

120

unit : mmunit : mm

Specifications
Model

Resolution Measuring Range (mm)(mm)( ) : Indicating Range*1

Indication
Error(μm)

Parallelism Measuring Force
(μm)
(N)

Contact Point Form /Anvil Form(mm)Weight(g)

Suggested
List Price

SMD-540A

0.01

12

±20

5

1.0 or less

φ10 Flat

290

41,800 JPY

SMD-565A

0.001

15(12)±3

3

1.5 or less

φ10 Flat

470

71,170 JPY

SMD-550A

0.01

12

±20

5

1.0 or less

φ10 Flat

44,440 JPY

440*1 Indicating value in ( ) is a measuring range of digital sensor.

SMD-565A-L

SMD-130A·Long stroke 25mm
·Resolution 0.01mm·Measuring Range 50mm
(Lifting of anvil)·Ceramic Contact Point
and Anvil

Dimensions:SMD-565A-L

Dimensions:SMD-130A

120

18.5

28

φ10

25
Lifting of
anvil

74

30

φ10

3

6.5

66.9

(219)

38.1

·Conversional type
without Anvil Adjustment.
·Resolution 0.001mm·Measuring Range 12mm
·Ceramic Contact Point and Anvil

7.5

45

72.5
unit : mm

95
unit : mm

Specifications
Model

Resolution(mm)

SMD-565A-L

0.001

SMD-130A

0.01

Measuring Range (mm)( ) : Indicating Range*1
1250(25)

Indication
Error(μm)

Parallelism(μm)

Measuring
Force(N)Contact Point Form(mm)

Anvil Form
(mm)

Weight(g)

Suggested
List Price±33

1.5 or less

φ10 Flat

φ10 Flat

415

58,080 JPY±205

2.0 or less

φ10 Flat

φ10 Flat

610

96,800 JPY*1 Indicating value in ( ) is a measuring range of digital sensor.

71

High precision measurement for papers, films andparts thickness.
φ45mm Mini-Indicator.
IP67.
One-touch data sending to PC by built-in Bluetooth.
Data management by SmartMeasure ® Lite (free).

Bluetooth Digital Thickness Gauge

17

43

9

Dimensions

6.5

φ10

SSS-5403

30.5

Ultra Light 176g

101

Features(11.2)

Ultra Light Compact SeriesNEW(95.5)

Thickness Gauge

Bluetooth Digital Thickness Gauge

30

14

7

72.5

unit : mm

Specifications
Model

Resolution(mm)

Measuring Indication
Range(mm) Error(μm)

Standard
Contact
Point(mm)Anvil Form(mm)

Insertion
Depth(mm)Weight(g)

Suggested
List Price

1.2 or less

φ3.2 Ball

φ10 Flat

φ10 Flat30

176

87,800 JPY

1.2 or less

φ3.2 Ball

φ10 Flat

φ5 Flat

φ3.2 Ball30

176

87,800 JPY

Measuring
Force(N)SSS-540

0.01

12

±20

1.2 or lessSSS-540-LW

0.01

12

±20

1.2 or lessSSS-540-LS
SSS-540-3A

0.01

12

0.01

12

±20
±20

30

φ5 Flat176

30

87,800 JPY

176

87,800 JPY

Contact point and Anvil for Thickness Gauge
Standard TypeContact
Point

LS TypeContact
Point

φ10

LW TypeContact
Point

Sφ3.2
φ10

Anvil

Anvil

Anvil

3A TypeContact
Point

Sφ3.2
Sφ3.2

φ5

Anvil

φ10 Flat : Ceramic φ3.2 Ball : Ball φ5 Flat : SK4 Hardened

17(11.2)

9

φ10

15

25

SSS-550

18

33

3Insertion Depth 120mm

110

(104)

43

Dimensions

7

14

120

162
unit : mm

Specifications
Model

Measuring Indication
Range(mm) Error(μm)

Standard
Contact
Point(mm)Anvil
Form(mm)

Insertion
Depth(mm)Weight(g)

Suggested
List Price

1.2 or less

φ3.2 Ball

φ10 Flat

φ10 Flat120

344

90,100 JPY

1.2 or less

φ3.2 Ball

φ10 Flat

φ5 Flat

φ3.2 Ball

120

344

90,100 JPY

Measuring
Force(N)SSS-550

0.01

12

±20

1.2 or lessSSS-550-LW

0.01

12

±20

1.2 or lessSSS-550-LS
SSS-550-3A

72

Resolution
(mm)
0.01
0.01

12
12

±20
±20

φ5 Flat

120
120

344
344

90,100 JPY
90,100 JPY

FeaturesHigh precision measurement for papers, films and parts thickness.
High resolution 1/1000mm.
IP54.
One-touch data sending to PC by built-in Bluetooth.
Data management by SmartMeasure ® Lite (free).

25.4

101

23

6.5
3

SSS-5650

28

49.6

φ59.5

46.8

Dimensions

φ10

(φ15)

30

7.5

Bluetooth Digital Thickness Gauge

7.5

Thickness Gauge

Standard Series

72.5

unit : mm

Specifications
Model

Resolution(mm)

SSS-5650

0.001

Measuring Indication
Range(mm) Error(μm)
12Long Stroke Model±3

Features

Standard
Contact
Point(mm)Measuring
Force(N)
1.2 or lessAnvil Form(mm)

Insertion
Depth(mm)Weight(g)

Suggested
List Price

φ10 Flat30

380

102,800 JPY

φ10 Flat

High precision measurement for papers, films and parts thickness.
Measuring Range 50mm , Indication Range 25mm(Lifting of anvil).
IP54.
One-touch data sending to PC by built-in Bluetooth.
Data management by SmartMeasure ® Lite (free).

7.5

25.4

74
(28.5) 9.5
Anvil moving(25.5)
25mm(218.9)(195.9)

φ59.5

Dimensions

120

95

φ10

18.5

45

SSS-130
Specifications

unit : mm※Measuring Range : Indicating value in ( ) is a measuring range of digital sensor.

Model

Resolution(mm)

SSS-130

0.01

Measuring Indication
Range(mm) Error(μm)
50(25)±20

Measuring
Force(N)
1.2 or lessStandard
Contact
Point(mm)φ10 Flat

Anvil Form(mm)

Insertion
Depth(mm)Weight(g)

Suggested
List Price

φ10 Flat45

543

153,100 JPY

73

Constant Pressured Thickness Measuring Instrument

Thickness measuring method for tested piece for physicaltest such as rubber, heat plasticity Elastomer, plastic film,
cloth, textile,leathers are ruled in detail by JIS or ISO.
PG/PF series are digital type thickness measuring instrument
in compliance with these major standard.

Stand type (fixed type) and frame type (handy type) are
widely used for test & research dept., quality control dept.
and manufacturing dept.

• Wide range of line-up complying with various standard of the field are available.
• High accuracy digital type with weight type for all the versions realizing stable static load, which is
not got by analog type utilizing gears or springs.
• PG series uses micro-granite which is superior for abrasion resistance , chemical resistance ,impact resistance in addition to high unstriated for measurable table. It can avoid scratches andstains for metal.
• Stainless steel is used for contact point and anvil (excluding partial model). Acid resistance , alkali
proof, water resistance are improved.
• Power source is silver oxide batteries (SR-44) which is convenient to carry.
• This makes treating statistics of measured data possible with connected to optional printerSD-763P.
• Contact point and measuring pressure can be changed. (However, it is not equivalent to standard )• Please refer to page 25 for specifications of each model.

Features

PG Series
stand type

Dimensions:PG-01A~PG-20A

PG-20AJIS K 6250 Method A
φ24.8

175

PG-20 is the thickness measuring
instrument compliant with A Law formeasuring thickness which isstandardized in JIS K 6250 (ruled inphysical test method general rule ofrubber for vulcanized rubber andthermoplastic rubber.)This is sheet block compatible typewhich can measure both thickness oftest piece hardness IRHD below 35
and over 35 by this one unit.
Contact point is diameter 5mm andpressure can be changed by onlyattaching and detaching weights.

12

50

φD

100

150
unit : mm

PF Series
frame typePF-02A

Contact Point

PF-11A

Contact Point

Anvil

Anvil

Dimensions:PF-01A/PF-02A

Dimensions:PF-11A~18A

162

φD

φ5

30

(72.5)

unit : mm

30

3.2 4

(30.7)

φ32

110

(160)

PF-02:(130)PF-01:(126)PF-01:φ20
PF-02:φ25

74

70

(50)

Thickness Gauge

Constant Pressured Thickness Measuring Instrument

120

unit : mm

Model
PG-01A

PF-01A
PF-02AJIS

K6732-1996

PF-11A

PG-12A

PF-12A

PG-13A

PF-13A

PG-14A

PF-14A

PG-15A

PF-15A

PG-16A

PF-16aPG-17A

PF-17A

PG-18A

PG-20A

PF-18A−

Poly vinyl chloride films for agriculture

K6783-1994

Ethylene / vinyl acetate copolymer films for
agriculture

Z1702-1994

Polyethylene film for packaging

Z1709-1995
PG-11A

Measured Item

K6400-1-2004
K6402-1976
K6301-1995

K6404-2-3-1999
K6328-1999
K6250-2006
L1086-2007
L1086-2007
L1096-1999
L1018-1999
L1086-2007
L1096-1999

12

0.001

12

Heat shrinkable plastic films for packaging
Flexible polyurethane foam
Flexible polyurethane foam for garments
Vulcanized rubber
Rubber or plastics coated fabrics
Rubber coated fabrics
Rubber A method(less than 35 IRHD)
Fusible interlining fabrics (Nonwoven textile)
Fusible interlining fabrics (Ordinary textile)
Woven fabrics (Ordinary textile)

0.8N(80±5gf)
1.22±0.14N(125±15gf)

1226±147mN

Woven fabrics (Crinose textile)

K6550-1994

Leathers

PG

PF

116,160 JPY 84,700 JPY

φ5116,160 JPY 84,700 JPY(125±15gf)
0.01

12

0.363N(37gf)

φ35.7

0.363kPa(3.7gf/cm2) 105,270 JPY 70,180 JPY

0.01

12

0.785N(80gf)

φ5105,270 JPY 70,180 JPY

0.01

12

0.785N(80gf)

φ10

0.01

12

0.394N(40gf)

φ16

0.01

12

2.35N(240gf)

φ11.3

10±1kPa105,270 JPY 70,180 JPY

10±2kPa2kPa

105,270 JPY 70,180 JPY

23.6kPa

23.5kPa(240gf/cm2)
0.7kPa(7gf/cm )113,740 JPY 79,860 JPY

2Fusible interlining fabrics (Ordinary knitting fabric)
Man-made upper material of shoes

Suggested List Price

φ5

Knitted fabrics (Ordinary knit)

K6505-1995

0.01

12

0.343N(35gf)

φ25.2

0.01

12

3.854±0.098N(393gf±10gf)

φ10

0.01

13

0.431N(44±10gf)

φ5

0.01

13

Rubber A method (35 IRHD or more)

K6250-2006

K6250-2006

0.001

Contact
Point
Measured Pressureφd (Mm)Rubber A method for both(less than 35 IRHD, 35 IRHD or more)

3.85±0.1N(390gf±10gf)

0.196 ± 0.038N(20±3.9gf)
0.431±0.098N(44±10gf )φ5

0.7kPa105,270 JPY 70,180 JPY

49.03±1.177kPa(500±12gf/cm2)

113,740 JPY 79,860 JPY

0.7kPa(7gf/cm2)(35 IRHD and over)22±5kPa(2.24±0.51gf/mm2)

105,270 JPY 70,180 JPY(Less than 35 IRHD)
10±2kPa (1.02±0.20gf/mm2)(35 IRHD and over)22±5kPa (2.24±0.51gf/mm2)

Constant Pressured Thickness Measuring Instrument

PG-02A

Resolution Measuring
Load
Range
(Mm)
(Measuring Force)(Mm)

Reference Standard

Stand-type Frame-type

Thickness Gauge

Specifications

113,740 JPY

Aluminium alloy is used for material of contact point (including anvil) of PG-11A and PF-11A. Contact point for other model are all stainless steel.
PG-13 and PF-13 can be also used for IRHD below 35 of JIS K 6250 A law.

• PF series can be used bybeing fixed to stand (Option).

Mounting example
PF Series

F M -18

Constant pressured thickness measuring
instrument of the plate-cylindrical shape
test piece (JIS K 6250 method A)

Dimensions

70

Weight B
Weight A

Hardness 35 for IRHD below-or more of vulcanized rubber.
φ59

Contact
Point
Contact
Point form

62

(53)

175

φ24.8

unit : mm

Specifications
Model

Less than 35 IRHD 35 IRHD and over Graduatuon Measuring Contact Point Suggested
Form
List Price
Range

PGM-20-510±2kPa(20gf)

22±5kPa(44gf)

0.01mm25mm

φ5mm

96,800 JPY

PGM-20-810±2kPa(51gf)

22±5kPa(113gf)

0.01mm25mm

φ8mm

96,800 JPY

75

Standard Model

NEW

Wide range of line-up by JIS or ISO.
HIgh accuracy digital type with weight type for stable static load.
Uses micro-granite which is superior for abrasion resistance,
chemical resistance and impact resistance in addition to high unstriated.
Stainless steel is used for Contact point and Anvil.
One-touch data sending to PC by built-in Bluetooth.
Data management by SmartMeasure ® Lite (free).

Features

Bluetooth Digital Constant Presured Thickness Gauge

Dimensions
φ24.8

50

175

(φ20 )

φ59.5

Thickness Gauge

Bluetooth Digital Constant Pressured Thickness Gauge

φ5
62

70

150

100

SSPG-0125

unit : mm

Specifications
Reference Standard

Model
JIS
SSPG-01
SSPG-02

K6732-1996

SSPG-12SSPG-13
SSPG-14
SSPG-15
SSPG-16

SSPG-17

SSPG-18

K6404-2-3-1999
K6328-1999
K6250-2006
L1086-2007
L1086-2007

12

0.363N(37gf)

φ35.7

Vulcanized rubber

0.01

12

0.785N(80gf)

φ5

0.01

12

0.785N(80gf)

φ10

0.01

12

0.394N(40gf)

φ16

0.01

12

2.35N(240gf)

φ11.3

Rubber or plastics coated fabrics
Rubber coated fabrics
Rubber A method(less than 35 IRHD)
Fusible interlining fabrics (Nonwoven textile)
Fusible interlining fabrics (Ordinary textile)

Fusible interlining fabrics (Ordinary knitting fabric)

K6550-1994
K6250-2006

1226±147mN(125±15gf)

Knitted fabrics (Ordinary knit)

0.01

12

Woven fabrics (Crinose textile)
Man-made upper material of shoes
Leathers
Rubber A method (35 IRHD or more)

0.01

12

0.01

13

*) Product Weight = nealy 3.7 kg (Stand and Indicator etc. weight) + weight for Load76

153,700 JPY

0.01

L1086-2007
K6505-1995

φ51.22±0.14(125±15gf)

Measured Pressure

Suggested List
Price

Flexible polyurethane foam

Woven fabrics (Ordinary textile)

L1096-1999

153,700 JPY

Flexible polyurethane foam for garments

Heat shrinkable plastic films for packaging

L1096-1999
L1018-1999

φ512Polyethylene film for packaging

K6301-1995

0.8N(80±5gf)

0.001

Z1702-1994

K6402-1976

ContactPointΦd (Mm)12Ethylene / vinyl acetate copolymer films for
agriculture

K6400-1-2004

Load(Measuring Force)

0.001

Poly vinyl chloride films for agriculture

K6783-1994
Z1709-1995

SSPG-11

Measured Item

Resolution Measuring
Range
(Mm)
(Mm)

0.343N(35gf)
3.854±0.098N(393±10gf)
3.85±0.1N(390±10gf)
0.431N(44±10gf)

φ25.2

φ10

φ5

0.363kPa(3.7gf/cm²)

153,700 JPY
153,700 JPY

10±1kPa

153,700 JPY

10±2kPa

153,700 JPY

23.6kPa

165,900 JPY

2kPa

23.5kPa(240gf/cm²)
0.7kPa(7gf/cm²)
0.7kPa

0.7kPa(7gf/cm²)
49.03±1.177kPa(500±12gf/cm²)

153,700 JPY
165,900 JPY
153,700 JPY
153,700 JPY
153,700 JPY
165,900 JPY
165,900 JPY(35 IRHD and over)22±5kPa(2.24±0.51gf/mm²)

153,700 JPY

TypeLS t ype

C ontac t poi nt iss pheri c al , the a n v il
i s fl at.

SM-112LS
SM-528LS
SM-114LS
SM-124LS
SM-130LS
SMD-540S2-LS
SMD-550S2-LS
SM-112LW
SM-528LW
SM-114LW
SM-124LW
SM-130LW
SM-1201LW
SMD-540S2-LW
SMD-550S2-LW

Sφ3.2
φ10

Anvil

LW t ype

Contact
Point

C ontac t poi nt. a n v il
w i th s pheri c al .

Sφ3.2
Sφ3.2

Anvil

3A typeContact
Point

U pper and l ow e r
both φ 5 fl at.

2.5

φ5
Anvil

NE needle typeU pper and l ow e r
w i th needl e ty p e .

Contact
Point

2.5

φ2
Anvil

BL blade type

2.5

U pper and l ow e r
w i th bl ade ty pe .

Contact
Point
0.5
Anvil

4

U pper and l ow e r
w i th bl ade ty pe .
30°

Contact
Point
0.5

4

Anvil

2.5

KN blade typeLD flat disk type

Upper and lower
with a flat disk type. Contact
Point

φ30

Anvil

Graduation Measuring(mm)
Range(mm)Contact Point(mm)

Anbil Form(mm)

φ10 Flat
φ3.2 Ball
φ10 Flat
φ3.2 Ball
φ10 Flat
φ3.2 Ball
φ10 Flat
φ3.2 Ball
φ10 Flat
φ3.2 Ball
φ10 Flat
φ3.2 Ball
φ10 Flat
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3 Ball(Carbide) φ3 Ball(Carbide)
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball
φ3.2 Ball

0.01
0.01
0.01
0.01
0.01
0.01
0.01
0.01
0.01
0.01
0.01
0.01
0.001
0.01
0.01

10
20
10
20
50
12
12
10
20
10
20
50
10
12
12

SM-112-3A

0.01

10

φ5 Flat

φ5 Flat

SM-528-3A

0.01

20

φ5 Flat

φ5 Flat

SMD-540S2-3A

0.01

12

φ5 Flat

φ5 Flat

SMD-550S2-3A

0.01

12

φ5 Flat

φ5 FlatSM-112NE

0.01

10

φ2 Flat

φ2 Flat

SM-528NE

0.01

20

φ2 Flat

φ2 FlatSM-114NE

0.01

10

φ2 Flat

φ2 Flat

SMD-540S2-NE

0.01

12

φ2 Flat

φ2 Flat

SMD-550S2-NE

0.01

12

φ2 Flat

φ2 FlatSM-112BL0.01

7

t0.5/w4

t0.5/w4

SM-528BL0.01

17

t0.5/w4

t0.5/w4

SM-114BL

0.01

7

t0.5/w4

t0.5/w4

SMD-540S2-BL

0.01

10

t0.5/w4

t0.5/w4

SMD-550S2-BL

0.01

10

t0.5/w4

t0.5/w4

SM-112KN

0.01

7

t0.5/w4/30°

t0.5/w4/30°

SM-528KN

0.01

17t0.5/w4/30°

t0.5/w4/30°

SM-114KN

0.01

7

t0.5/w4/30°

t0.5/w4/30°

SMD-540S2-KN

0.01

10

t0.5/w4/30°

t0.5/w4/30°

SMD-550S2-KN

0.01

10

t0.5/w4/30°

t0.5/w4/30°

SM-112LD

0.01

10

φ30

φ30

SM-528LD

0.01

20

φ30

φ30

SM-114LD

0.01

10

φ30

φ30

SMD-540S2-LD

0.01

12

φ30

φ30

SMD-550S2-LD

0.01

12

φ30

φ30

Thickness Gauge Special Product List

Contact
Point

Adaptation models

Thickness Gauge

Thickness Gauge Special Product List

●SM is Analog Type, SMD is Digital type.

77

Thickness Gauge

Parts & Accessories
Contact Point, Anvil of Symbol and Shapes
Standard typeLS typeParts & Accessories

Contact Point

Contact Point
φ10

Anvil

unit : mm

LW typeSφ3.2

3A typeContact Point

φ10
Anvil

Anvil

Sφ3.2
Sφ3.2

Contact Point
φ5
Anvil

FE type
φ10 flat steel
SK4 Hardened
SK4 Hardened

AT type
φ10 flat steel
Ceramic

φ3.2 Steel Ball

φ3.2 Steel Ball

Ceramic

Ceramic

φ3.2 Steel Ball

SK4 Hardened

Specialized Coating

SK4 Hardened

Anvil impossible replaces because of the adhesion.

Special order product of Dial Thickness Gauge / Digital Thickness Gauge
1. Instruct dimension and shape of anvil and contact point by referring to the figure in the right and P77.
2. Instruct necessary measuring range.
3. In case that there is direction like blade type, instruct “parallel” or “right angle” based on graduation face as front facestandard.
4. In case of requesting shape of anvil and contact point rather than figure in the right or change of measuring force, pleasecontact our nearest branch for you.

78

NE (needle) typeBL (blade) type

KN (knife edge) type

LD (large diameter flat) typeThis is top and bottom
needle type(cylinder).
Instruct diameter

This is top and bottom
blade type(blade).Instructwidth and thickness.

This is top and bottom
knife edge type. Instructwidth and angle.

This is top and bottom
disc type(cylinder).
Instruct diameter.