Precision Hydrogen 1200
Hydrogen Generator for GC
Part Number : See Reverse

Service Kit : See Reverse

Your local gas generation partner

Description
The Precision Hydrogen generators are designed to provide thegas needed for detectors requiring Hydrogen fuel gas, such as FID
and FPD. One generator is capable of supplying multiple detectors,
and are available in various flow rates to ideally suit individual
customers’ needs.
These generators utilize a Proton Exchange Membrane (PEM) togenerate Hydrogen gas from deionized water, with a pressure
swing adsorption (PSA) dryer used to remove residual moisturefrom the gas.
Precision Hydrogen gas generators come with various safety
features as standard, giving you complete peace of mind in thelaboratory and a far safer, dependable and more convenient
alternative to cylinder gas.

Applications)

Detector Gas)

Fuel Gas

Key Features

540 mm380 mm406 mm)

Suitable for fuel gas at standard detection limits)

99.9995% Purity)

Internal leak detection with automatic shutdownfeatures)

Automatic loading pump as standard)

Simple maintenance, limited to replacing de-ionizer)

Short and easy start-up and shutdown procedures)Compact, space-saving modular design)

Creates hydrogen on demand, minimal storage ofhydrogen in the system)

Capable of supporting up to 40 FID detectors

Precision Hydrogen 1200 shown in a typical Precision stack.

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Hydrogen 1200 Rev 8 RSID 2552 EN —10/09/2024

Technical Specifications

Hydrogen, 1200cc

Max Flow Rate*

1200cc/min

Max Pressure

100psi / 6.9 bar

Purity99.9995%

Gas Outlets

1 x 1/8” Swagelok compression fitting

Water Purity Requirement<1.0μ Siemens/cm OR >1 Mohm-cm

Water Consumption

up to 2.1 L/day

Operating Temperature

10°C - 35°C / 50°F - 95°F

Electrical Requirements

110-230V 50/60HZ 10A Max

Power Consumption

600 Watts

Heat Output

up to 1000 BTU/hr

Generator Dimensions (WxDxH)

38.0x 54.0 x 40.6cm / 14.9 x 21.2 x 15.9”

Generator Weight

31.5Kg (69.5 lbs)

Noise Level

Silent in operation* All flow rates are stated in sccm at 273.15K and 1.01bar

Ordering Information
Part Number

3300843

Annual Service
Standard Maintenance Plan

www.peakscientific.com/ordering

Complete Maintenance Plan

Accessories

Water Bottle 4L

Water Bottle 8LWater Bottle

10-9016

10-9017

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UKTel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Hydrogen 1200 Rev 8 RSID 2552 EN —10/09/2024