Solaris XE
Nitrogen generator for LC-MS or multiple ELSDs
Part Number : 99-9999

Service Kit : 99-9999

Your local gas generation partner

Description
Peak Scientific’s Solaris XE generator has been engineered using
membrane technology to meet the gas delivery requirements
for labs using LC-MS, Compact MS or multiple ELSDs. Solaris XE
can operate with varying flow rates (up to 35L/min of high purity
nitrogen), purity (95-99.5%) and outlet pressure can be adjusted
down from 116psi, offering a flexible solution for a variety ofapplications.
The Solaris XE range has been engineered to provide nitrogen tolaboratories that utilize an external source of compressed air, whichmeets a minimum quality grade of ISO 8573-1:2010 Class 1.4.1,and its compact, space-saving chassis allows it to be placed on abenchtop, taking up minimal space in the lab.

Applications
LC-MS, Compact MS, Multiple ELSDs, TLD readers

Key Features
Nitrogen up to 35L/min, up to 116 psi @ 95-99.5% purity
Variable outlet flow, pressure and purity
Compact, stackable* system maximising the use ofvaluable laboratory floor or bench space
Colour indicative LED lighting for easy status awareness
Manufactured and tested to highest spec in UK, CE/FCC
certified
Safer and more convenient than pressurized cylinders,
dewars or bulk storage
12 month on-site comprehensive warranty
Backed by [Peak Protected] global on premise servicing
and technical support*units stacked on top of Solaris must not exceed 100kg/220.4lbs

Copyright © Peak Scientific Instruments Ltd 2022 — Solaris XE Data sheet Rev 4 RSID 2297 EN —01/02/2022

Technical Specifications

Solaris XE

Max Flow Rateup to 35 L/min

Max Pressureup to 116 psi / 8 bar(user adjustable)

Purity95-99.5 %*Gas Outlets

1 x 1/4 “ BSPP

Min/Max Air Inlet Pressure

80 psi/5.5 bar - 145 psi/10 bar

Min Inlet Air Quality

ISO8573 - 1:2010 Class 1.4.1

Contaminants

Phthalate & BHT Free

Suspended Liquids

None

Operating Temperature

5°C - 30°C / 41°F - °F

Electrical Requirements

Universal100 / 230 VAC / 50 / 60 Hz / 0.1 A

Power Consumption

6.9 Watts

Heat Output

25 BTU/Hr

Generator Dimensions (HxWxD)

156 x 343 x 650 mm
6.1 x 13.5 x 25.6’’

Generator Weight

15.1 kg / 33.3 lbs

Max Weight Bearing Capability100 kg / 220.5 lbs

Solaris Compressor compatible

NO*depending on inlet air pressure and outlet flow

Ordering Information
Part Number

3300671

Annual Service

www.peakscientific.com/ordering

FPPM Maintenance Plan

3300671-FPPM

Complete Maintenance Plan

3300671-CSP

Supported Maintenance Plan

3300671-SMP

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2022 Peak Scientific - Rev - Solaris XE Data sheet Rev 4 RSID 2297 EN - 01/02/22