Other Precision Measuring Instrument
Dial Tension Gauge• These are mechanical tension gauges displaying in Newton, which can measure precise and moment force like tensile force or compression force• Miniature bearing is adopted to contact point bearing. and it can obtain stable indication accuracy.
• It can be used for the wide range of usage for contact force such as micro-switch, relay, valve, and measuring force of dial indicator, and springforce of precise spring, and pressing force of key-board.
• There are 2 types of standard type which sustains momentary highest value, and peak pointer type .
Dimensions

01.5
33.5(43.5)

0.8
Fulcrum

490(100)

Contact PointHand

Peak Pointer55

φ6Standard Type

Peak Pointer Type

DTN-50

DTN-100G

Graduation 0.02N
Measuring Range 0.06N〜0.5N

格0Measuring holeφ13

Both direction isapplicable39.5(49.5)

9.3

Graduation 0.05N
Measuring Range 0.1N〜1N

Specifications: Newton Force

19(5)

47

unit：mm( ): DTN-5、DTN-10、DTN-10G

Specifications: Gram Force

Model

Graduation

Measuring Range

Dial Reading

WeightModel

Graduation

Measuring Range

Dial Reading

WeightDTN-52mN

6mN〜50mN

6-50-654g

DT-50.2gf

0.6gf〜5gf

0.6-5-0.654g

DTN-105mN

10mN〜100mN

10-100-1054g

DT-100.5gf

1gf〜10gf

1-10-154g

Standard

DTN-3010mN

30mN〜300mN

30-300-3054g

Standard

DT-30

1gf

3gf〜30gf

3-30-354g

TypeDTN-50

0.02N

0.06N〜0.5N

0.06-0.5-0.06

54g

TypeDT-50

2gf

6gf〜50gf

6-50-654g

DTN-100

0.05N

0.1N〜1N

0.1-1-0.1

54g

DT-1005gf

10gf〜100gf

10-100-1054g

DTN-150

0.05N

0.15N〜1.5N

0.15-1.5-0.1554g

DT-1505gf

15gf〜150gf

15-150-1554g

格DTN-300

0.1N

0.3N〜3N

0.3-3-0.3

54g

DT-300

10gf

30gf〜300gf

30-300-30

54g

40DTN-500

0.2N

0.6〜5N

0.6-5-0.6

54g

DT-500

20gf

60gf〜500gf

60-500-60

54g

Model

Graduation

Measuring Range

Dial Reading

WeightModel

Graduation

Measuring Range

Dial Reading

Weight

DTN-10G

5mN

10mN〜100mN

10-100-10

56g

DT-10G

0.5gf

1gf〜10gf

1-10-156g

DTN-30G

10mN

30mN〜300mN

30-300-30

56g

DT-30G

1gf

3gf〜30gf

3-30-356g

0Type

000005050Type50TypeType50

Peak

DTN-50G

0.02N

0.06N〜0.5N

0.06-0.5-0.06

56g

Peak

DT-50G

2gf

6gf〜50gf

6-50-656g

50Pointer

DTN-100G

0.05N

0.1N〜1N

0.1-1-0.1

56g

Pointer

DT-100G

5gf

10gf〜100gf

10-100-1056g

40DTN-150G

0.05N

0.15N〜1.5N

0.15-1.5-0.15

56g

DT-150G

5gf

15gf〜150gf

15-150-1556g

す）

DTN-300G

0.1N

0.3N〜3N

0.3-3-0.3

56g

DT-300G

10gf

30gf〜300gf

30-300-30

56g

DTN-500G

0.2N

0.6N〜5N

DT-500G

20gf

60gf〜500gf

56g
0.6-5-0.6
Accuracy＝±1/2graduation

Push Pull Gauge• It is a multi-purpose dynamometer to use the Y head or L hooks.

Dimensions:
PPN-705-3/5/10

56g
60-500-60
Accuracy＝±1/2graduation

Dimensions:
PPN-705-20
PUSH

25.4

Y-HeadHand

Hand

Y-Head(175)

Peak
Pointer

φ15

79.2

Outer Dial

85.3

(177)

Peak
Pointer

φ40

φ40(55)

51.7

31.7

PUSH

Outer Dial
φ20

Graduation 0.05N
Measuring Range 0.35N〜3N

PPN-705-5

Graduation 0.05N
Measuring Range 0.4N〜5N

Specifications: Newton Force

PPN-705-10(30)(40)

PPN-705-20

Graduation 0.1N
Measuring Range 0.5N〜10N

J-HookJ-HookGraduation 0.2N
Measuring Range 0.6N〜20N

PULL

unit : mm

PULL

unit : mm

Specifications: Gram Force

Model

Graduation (N)Measuring Range

Dial Reading

Weight (g)PPN-705-3

0.05

0.35N〜3N

0.35-1.5-385Standard

PPN-705-5

0.05

0.4N〜5N

0.4-2.5-585

Type

PPN-705-10

0.1

0.5N〜10N

0.5-5.0-10

85

PPN-705-20

0.2

0.6N〜20N

124
0.6-10-20
Accuracy＝±1graduation

Type(31)

PPN-705-3(41)L-HookSupplied asstandard accessory

TypeStandardType

Model

Graduation (gf)Measuring Range (gf)

Dial Reading

Weight (g)PP-705-300

535〜300

35-150-30085

PP-705-500

540〜500

40-250-50085

PP-705-1000

1050〜1,000

50-500-1000

85

Accuracy＝±1graduation

47

Other Precision Measuring Instrument

Crankshaft Deflection Gauge• Exclusive instrument to measure deflections of engine crank of automobiles and ships etc.

Dimensions

C33

φ33

CSDG-A

Extension Rods
6441.3L

Graduation 0.01mm
Measuring Range 100〜230mm
27（37）

Specifications
ModelGraduation

Measuring

Indicator Error

Weight（mm）

Range（mm）

（µm）

（g）

0.01

100〜230±15300

CSDG-A39.5A

φ8φ87.5

B（31.5）

Dimensions Table(Extension Rods)
NO.
ABCL
3580120

unit：mm

Measuring Range
100〜145mm
145〜190mm
185〜230mm
unit：mm

Crankshaft Deflection Gauge with magnet• Small type which can be fixed to measuring position by magnet.
Dimensions

DE

φF

Indicating Range Amm

TM-106YS

Graduation 0.01mm
Measuring Range 65〜82mm

CTM-104YS

Graduation 0.01mm
Measuring Range 99〜113mm

Specifications
Model48

Graduation

Measuring Range

Indicating Range of Dial

Dial Gauge ErrorHolding

Weight（mm）（mm）

Indicator (mm)（µm）

Power（N）（g）

TM-104YS

0.01

65〜82

4±15100230

TM-106YS

0.01

99〜1135±15100

330

φB

Dimensions TableModel
TM-104YS
TM-106YSA
45B
3532unit：mm
C
2232

DE
F
4.5 68.5 35
7 102.3 47
unit：mmOther Precision Measuring Instrument

Small Bench Center• Suitable to measure eccentricity volume (swing) of various precision shafts, round manufactured work piece, precision gears and pinions.
• Convex and concave of center can be easily changed according to shape of work piece. Center can be shifted.
• Standard type (graduation 0.01mm) and precision type (graduation 0.001mm) are available.
Dimensions

Dial Indicator

FM-560°

unit：mm

Measuring

Dial indicator Error

Measuring ForceApplicable Dimension

Weight(mm)Range (mm)

(μm)

(N)

Diameter

Width(kg)FM-5 Standard Type0.01

10±151.4 or less

Up to φ60mm

Up to 60mm

2FM-5 High Precision Type0.001

1±51.5 or less

Up to φ60mm

Up to 60mm

2（4.8）
120°
60°

Graduation

φ1φ8

Convex and concave
center measuring

60°

φ6.4

Dimensions : Center Pole
φ4Concave Center SideSpecifications
Model（78）（180）

32（73）

Both concave
center measuring
Convex Center Side

0〜75

358020Both convex
center measuring

Centerφ20（19）（120）

Measurement example for swingof round bar.
WorkMAX.67
MIN.229Standard type

unit：mm

Flow Meter¡This is a dial indicator and jig set for the suction (expansion/ contraction) measuring.

Dimensions

φ5571687

150

Graduation 0.1mm
Measuring Range 25mm

Specifications
Graduation (mm)Measuring Range (mm)Gauge Error

Dial Reading

716

0.1

25±60µm

0-5-10

Short Hand Weight (g)non8Model260g

φ13φ19Calibration Tester

unit：mm

Applicable gauges and industrial standards
0.01mm Dial Indicators• FM-10 is a calibration device to check accuracy of dial indicators, test indicators, back plunger
type dial indicators and bore gauges. Accuracy of various gauges mentioned in the following list
can be measured according to combination of various units. Optional pole (ZY-095) is neededfor 400mm-600mm measuring range bore gauges.

JIS B 7503

0.01mm Dial Indicators (20mm)

JAMS 2001

Small Dial Indicators

JAMS 2001

0.001mm Dial IndicatorsTest Indicators

Back Plunger Type Dial Indicators
Bore Gauges

Bore Gauges for small holes

JIS B 7503
JIS B 7533
–JIS B 7515

JAMS 2009

FM-10Graduation 0.001mm
Measuring Range 25mm
Weight 9kg
Used Micrometer Head
Pitch 0.5mm
Range 25mm
Sleeve graduation 500
Instrumental error ±1μm
It does not include calibration certificate.

For Dial Indicators

For Dial Test Indicators

For Back Plunger Dial Indicators

For Bore Gauges49