Alleviate fatigue caused bystanding for a long timeConductive Anti-Fatigue Mat
This conductive mat can be used on the ESD preventive floor.
Conductivity rating : Ｒ≦1.0×109 Ω
EDP No. 621489

EDP No. 621488

For indoor use only
Uniform and stable conductive performance
Thickness avoids warping
Gray color for easy maintenance600447※Grounding wire is not attached.

600

900

SPECIFICATIONSSPECIFICATIONS

AF-45
W600mm × L447mm × t15mm
Gray
Ｒ ≦ 1.0×109 Ω2.0kg
NBR

Model No.
Size
Color
Conductivity
WeightMaterial

Model No.
Size
Color
Conductivity
WeightMaterialAF-90
W900mm × L600mm × t15mm
GrayＲ ≦ 1.0×109 Ω4.2kg
NBR

Uniform and stable conductive performance
Matte surface ﬁnish, reducing reﬂections for less eye-fatigue
Heat resistant to dropping solder

EPA Working Mat

Conductive Rubber Mat
Controlling electro conductivity
between 106 and 107 Ω
without influence fromthe working environment.
Ideal for workbenches requiring
static electricity measures.

Conductivity

Material

10 ≦R≦10 Ω

NBR + SBR

6Surface layer(static electricity-diffusing rubber)(conductive rubber)

COMMON SPECIFICATIONS
Model No.

EDP No. 621461

Easy checking of surface resistance on floor or mat

Conductivity

10 6≦R≦10 7Ω

10 6≦R≦10 7Ω

Weight35kg/roll

35kg/rollMaterial

NBR＋SBR

NBR＋SBR

All items necessary for 4 types of measurement in one package
Can also measure temperature and humidity (the values are for reference)
Applied voltage selector switch enables to measure either the range of1.0x103〜105 Ω or of 1.0x106〜1012 Ω.
Measurement

Measurement

Resistance between a ground anda points on a surface

Probe (outer diameter: about Φ66 mm, about 2.3 kg): 2

SPECIFICATIONS

101.5Warning

Measurement Measurement

Surfaceresistivity

Directive

Volumeresistance

Conformity standards

IEC61340-4-1,

Curled cord for measurement for the point-to-point resistanceand resistance to ground (banana plug/mono plug): 2

Measurement range

103〜1012Ω
（Ω/sq.）

Applied voltage

10V (R＜1.0×10 6Ω or Ω/sq.) , 100V (R≧1.0×10 6Ω or Ω/sq.)

Crocodile clip: 1, Large crocodile clip: 1

Display

Triple-digit LCD (□.□□ e□□) ex.: 3.14 e10 → 3.14×1010

Stainless plate (230 mm x 142 mm): 1

Power source

9V battery × 1 (Type: 006P, recommended alkaline battery)

Stainless plate: 1

L×W×D

190（L）× 101.5 (W) × 54.7 (H) [mm]Carrying case: 19V battery (006P type: manganese dry battery): 1Weight

Body: around 340g (including a battery),gross weight: around 6.2kg (including the carrying case)O p erating tem p eratu rea n d hu midit y

0℃〜37℃, 10%〜90%RH

MaterialBody case: ABS resin

Curled cord for a body earth (banana plugs at both ends): 154.7

GreenGreen
W1.0m×L10m×t2.0mm

ACCESORIES

190

Lightgreen
Light green

Resistance between
2 points on a surface

EDP No. 621604
66

SG-100LG-100W1.0m×L10m×t2.0mm

Size

Surface Resistance Checker

124.3

EPABacking

EDP No. 621464

ROLL

7

STRUCTURE

Color10m

DirectiveApplications

Designed for use
with work benches※The accompanying battery is a sample.
It may have run down soon after purchase.
Thank you for getting a new battery by yourself.

JIS-C2170

Manuals●For safety purposes, read the instruction manual carefully before using the unit. ●Do not use this product in an explosion-proof area. ●A high voltage is applied on this product.
Make sure that water, oil, solvents, etc., do not come in contact. ●Avoid dew condensation as it can result in electric shock or product damage.
●Keep away metal objects such as tools or needles, or body parts such as fingers, hands or face from the needle electrode because a high voltage is applied on the needle electrode.

●For improvements, the product specifications, size, price and other information
may be subject to change without prior notice.

16080021.02