Stand
Theses are stands which can make precision measurement of work piece by mounting Dial Indicator or Digital Indicator. As to natural granite(micro-granite) base, its organization is fine particle and work piece slides smoothly without ringing. Stable and high flatness accuracy within
0.001mm / 200mm square is also characteristics. Please select referring to size of base or fine adjustment device from various lineup.

Granite Comparator Stand• Natural micro-granite is used for all granite comparator stands.
• Flatness accuracy : within 0.001mm / 200mm square.
• Sliding smoothly and objects to be measured are not damaged. It issuperior for anti-abrasion.
• The rigid granite base is free from burrs and pileups.
• It is easy for maintenance as base never suffers from rust.

USG-10

USG-24

Specimen table size
100×150×50mm

USG-29

USG-30Specimen table size
150×200×50mm

Specimen table size
150×200×50mm

Dimensions

Dimensions

Dimensions
Micro-Motion
micro-motion Range5.0
range5.0

Bushφ8240AB

USG-10,14,18

Dimensions

MAX.167
MIN.3758
340〜200C

A:150

B

USG-20,24,28

unit : mm

80505050A

Dimensions

Alternativefor left and right（φ25）

（φ25）

304

70
D70

Micro-Motion
micro-motion
Range2.0
range2.0CC

Alternative
for left and right（φ25）

Bushφ8Bushφ8D

Alternative
for left and rightSpecimen table size
150×150×50mm

USG-29unit : mm

MAX16734

B:200

unit : mm

MIN37

φ25.8

280

φ25.8

φ45D

D280

25 28

Bushφ8φ29.8

ACC

φ29.8

USG-30ABunit : mm

USG-31,32

B

unit : mm

Specifications
Standard

Specimen Table Size
(A×B) (mm)100×150Height of Pole (C)
(mm)
175

Movable Range (D)
(mm)
0〜130

Weight(kg)3.5

USG-14Standard

150×150

1750〜130

4.9

USG-18

Standard

150×200

240

0〜200

5.8

USG-20Fine Adjustment

100×150

1750〜115

3.5

USG-24

Fine Adjustment

150×150

1750〜115

4.9

USG-28Fine Adjustment

150×200

240

0〜180

5.8

USG-29Precision Fine Adjustment

150×200

240

0〜200

Model

Type

USG-10

Model

Type

USG-30

Square Screw, Arm Slide
Square Screw, Arm Slide
Square Screw, Arm Slide

USG-31
USG-32

6.0

Specimen Table Size （A×B×C）（mm）
150×200×50

Movable Range (D)(mm)
53〜250Bush φE(mm)φ8Weight(kg)7.3

200×250×50

53〜250φ8

10.5

250×250×50

53〜250φ812.5For USG30~32, dependingon the measuring device inuse, the probe may notcontact to the base surface
since the effective movable
range is 53mm or more. Usean extension rod or the liketo extend.
Other combinations ofspecimen table and pole unit
than the above chart arealso possible.
The specimen table ispossible to be made up to500x500mm according to
the requested size.

55

StandUpright Stand• This can be easily used by only mounting Dial Indicator for comparison measurement.
• This small stand is suitable for measuring small parts etc.

US-22B

Specimen table size 80×80mm
Standard Type
Precision Grinding Cast Iron baseDimensions : US-16B

Specimen table size φ60mm
With fine adjustment device
φ60mm Ceramics baseDimensions : US-22B

102(φ25)

Alternativefor left and right

6580
φ60

148150unit : mm

100166unit : mmunit : mm

Specifications
Specimen Table Size （mm）

Movable Range (mm)Weight(kg)US-16B

Standard Type

80×80

0〜140

3.4

US-22B

With Micro-Motion Device

80×80

0〜110

3.5

US-25

Parallel Spring Type Precision Micro-Motion Device

φ600〜120

3.9ModelTypeHolder Stand for Thickness Gauge
Dimensions
204817φ34

MAX 15FM-18

M6×P1.0862722.5

approx.45° 25
9movable

φ72
φ92

unit : mmSpecifications
ModelFM-1856(φ25)

588016041158080

Micro-Motion
micro-Motion
range 2.0Range
2.0

175

MAX110(φ25)

Alternativefor left and rightBushφ840

MAX 140

175653880

Dimensions : US-25

Bushφ8Micro-motion
Micro-Motion
Range
5.0
range 5.0

Bushφ8
Alternative
for left and right

US-25

Specimen table size 80×80mm
With fine adjustment device
Precision Grinding Cast Iron base

MAX120US-16B

Size (mm)Weight(g)base φ92 , hight 86850

StandMagnetic Base• Magnetic base can hold a measuring device such as dial indicator.
• Adaptor (Internal diameter φ6mm) is supplied for holding testindicators on model : MB-1040 and MB-1050• MB-1040 / 1050 can also be fixed with dovetail of auto-clutch levertest.

• Dial indicator is mounted to MB-B with lug back and test indicator isclamped with stem (It can not be clamped with dovetail).
• MB-1040 / 1050 can not be clamped with dovetail.

MB-B

MB-1050

MB-1040

ExampleMagnetic Base with Test Indicator.
Dial indicator is also applied.

Dimensions: MB-1040

φ6Dimensions: MB-B

1Dovetail27φ8165

27 φ8175(Test Indicator)

51

Dovetail(Test Indicator)

10φ6Dimensions: MB-1050

φ12φ10Fine Adjustment Knob

56

Fine Adjustment Knob

11ON/OFF Knob

440ON/OFF Knob
OFF

6555

ON40unit : mm

605062unit : mmunit : mm

Specifications
Model

TypeMB-1040

One-touch, fine adjustment, small

320

6.0/8.0

0.5

MB-1050

One-touch, fine adjustment800

6.0/8.0

1.4Standard

800

4.5/6.5

1.5

MB-B

Holding Power (N) Mount Hole Diameter(mm)Weight (kg)57