取扱説明書

HANDHELDDATA LOGGER

TC-32K

写り防止ページ

はじめに
本書は、ハンドヘルドデータロガー TC-32K の操作方法や手順について説明しています。
本製品の全機能を生かし効率良く、正しい測定をしていただくためにも、本書をよくお読みになり、機能・操作を十分に理解され、ご使用いただくようお願いいたします。
本書はいつでもご使用いただけますよう、大切に保管してください。

■本書の見かた本書では、次のような記号を用いて重要事項の説明をしていますので、お読みください。
危険

この表示を無視して誤った取扱いをすると、人が死亡または重症を負う危険が想定される内容を示しています。

警告

この表示を無視して誤った取扱いをすると、人が傷害を負う危険が想定される内容を示しています。

注意

この表示を無視して誤った取扱いをすると、物的障害の発生が想定される内容を示しています。

注記

誤りやすい操作などについての注意や補足を示しています。

✍

本文の内容について理解を深める事柄や、知っておくと役に立つ情報を示しています。

❏本書の一部または全部を無断で転載、複製することは、固くお断りします。
❏本書の内容については、性能・機能の向上などにより予告なく変更することがあります。
❏本書の内容について、ご不明な点や誤り、記載もれなどお気付きのことがありましたら
最寄の営業所までご連絡ください。

本書記載の会社名、商品名は、各社の商標および登録商標です。
本書は、ソフトバージョン 1.2 に対応します。

安全上の注意
危険

引火性ガス、または引火性の蒸気のある場所で使用しないでください。引火することがあり大変危険です。

危険お客様による分解、改造などは感電や故障の原因になりますのでおやめください。

危険アース線は絶対にガス管につながないでください。また、アース線をつないだりはずしたりする時は、電源ケーブルを必ず先にはずしてください。火災、感電の危険があります。

警告

内部に粉塵などが入るとコネクタの接触不良や絶縁低下などの故障の原因になります。使用時だけでなく、保管のときも粉塵などが入らないようにしてください。

注意

仕様の温度範囲でお使いください。また、やむを得ず直射日光が当たる場合や、寒冷地でお使いになるときには、日よけや保温をしてください。

注意

相対湿度 85％以下でお使いください。雨のかかるところや、非常に湿度の高いとこ
ろでのご使用は、故障の原因になります。 内部に水が入った場合、あるいは冠水
した場合は、十分に乾燥してから電源を投入してください。正常に起動しない場合
は故障の可能性がありますので、弊社までご連絡ください。

注意

大型電動機・クレーン・変圧器や溶接機の近くに置いたり、配線を近づけたりしないでください。また、変電所やラジオの送信所の近くなど、電界が強い場所でのセンサの延長には、シールドなどの対策が施されたケーブルを使用してください。

注意

落雷・誘導雷の影響を受けることがあります。落雷のおそれがある場所には、落雷対策を施してください。また、不明な点は弊社までご相談ください。

注意

各接続ケーブルをつないだまま無理に引っ張らないでください。ケーブルが断線したり、コネクタが抜けることがあります。同様に、ケーブルのコネクタ部分に衝撃を加えたり、土、泥、水、油などに浸けないようにしてください。

注意

本器の電源電圧範囲は DC9V～18V です。それ以上の電圧で使用しますと、
故障の原因となりますので注意してください。＋極アース車で使用する場合は、
車体フレームと本器の筐体を接触させないでください。

目次
はじめに
安全上の注意目次

第１章 概要１.１ 概要 ································································································································································· 1 - 2
１.２ 特長 ································································································································································· 1 - 2
１.３ 各部の説明 ·················································································································································· 1 - 3
前面、側面 ············································································································································ 1 - 3
上面、底面、背面 ······························································································································· 1 - 4

第２章 準備
２.１ 使用上の注意 ············································································································································· 2 - 2
２.２ 電源について ···············································································································································
電池の入れ方 ······································································································································
電池での使用時間 ····························································································································
AC アダプタの接続 ····························································································································
電源の ON/OFF ·································································································································

2 - 3
2 - 3
2 - 4
2 - 5
2 - 5

２.３ 現場測定での注意 ···································································································································· 2 - 6
アースの接続 ······································································································································· 2 - 6
落雷対策 ··············································································································································· 2 - 6
２.４ 画面表示の概要 ············································································································································ 2 - 7
起動画面について ····························································································································· 2 - 7
画面構成について ····························································································································· 2 - 7
２.５ 操作の概要 ···················································································································································
操作体系 ···············································································································································
キースイッチ ·········································································································································
キーロック ··············································································································································

2 - 8
2 - 8
2 - 9
2 - 9

第３章 センサの接続
３.１ センサの接続 ·············································································································································· 3 - 2
入力端子台の使用方法 ·················································································································· 3 - 2
センサ別結線方法 ····························································································································· 3 - 3

第４章 モニタ表示と測定
４.１ モニタ画面の概要 ······································································································································ 4 - 2
４.２ モニタ表示 ····················································································································································
数値モニタ ············································································································································
波形モニタ ············································································································································
表示値の意味 ······································································································································
モニタの種類の選択 ·························································································································
波形モニタの設定 ······························································································································

4 - 2
4 - 2
4 - 3
4 - 3
4 - 4
4 - 4

４.３ 初期値の処理 ·············································································································································
イニシャルイン ·····································································································································
モニタチャンネルのイニシャルイン ······························································································
初期値の書換え ·································································································································

4 - 5
4 - 5
4 - 6
4 - 7

４.４ チャンネル設定 ··········································································································································· 4 - 8

測定モードについて ·························································································································· 4 - 8
測定モードの切替え ·························································································································· 4 - 8
シングルチャンネルモードのチャンネル設定 ·········································································· 4 - 9
４.５ 測定値の記録 ············································································································································· 4 -10
手動測定 ··············································································································································· 4 -10
自動測定 ··············································································································································· 4 -11
４.６ サブ LCD ······················································································································································· 4 -11

第５章 測定の設定
５.１ 測定の設定概要 ········································································································································· 5 - 2
５.２ センサモードの設定 ··································································································································
センサモードについて ·······················································································································
センサモードの設定 ··························································································································
グループ設定について ····················································································································

5 - 3
5 - 3
5 - 4
5 - 5

５.３ 係数・表示桁・単位 ···································································································································
パラメータについて ····························································································································
パラメータの設定例 ···························································································································
係数・表示桁・単位の設定 ·············································································································
Cap/RO からの係数設定 ················································································································
係数チェック ·········································································································································

5 - 6
5 - 6
5 - 6
5 - 7
5 - 9
5 -10

５.４ 熱電対基準接点の設定 ························································································································· 5 -11
５.５ TEDS センサ ··············································································································································· 5 -12
センサ設定の読込み ························································································································ 5 -12
読込んだ設定の反映 ························································································································ 5 -13
５.６ 測定モードの切替え ································································································································· 5 -13
５.７ メジャーとダイレクトの切替え ················································································································ 5 -14
５.８ 自動測定 ·······················································································································································
インターバル測定とは ······················································································································
インターバルタイマの設定 ··············································································································
リアルタイムスタートの設定 ···········································································································
Goto ステップとは ······························································································································
Goto ステップの設定 ························································································································
インターバル測定の開始/停止 ····································································································
スリープ機能 ········································································································································

5 -15
5 -15
5 -15
5 -17
5 -18
5 -19
5 -20
5 -21

５.９ 各種チェック ·················································································································································
絶縁チェック ·········································································································································
抵抗チェック ·········································································································································
ばらつきチェック ··································································································································
リード線抵抗チェック ·························································································································
ブリッジ出力チェック ··························································································································
係数設定チェック ································································································································

5 -22
5 -22
5 -22
5 -23
5 -23
5 -23
5 -24

５.１０ 測定補助設定 ··········································································································································
シンプルメジャーの設定 ··················································································································
Comet の設定 ·····································································································································
測定環境の電源周波数 ··················································································································
イニシャルインの許可 ······················································································································
バーンアウトチェックの設定 ···········································································································

5 -24
5 -24
5 -26
5 -26
5 -27
5 -27

第６章 記録の設定
６.１ 記録の設定概要 ········································································································································· 6 - 2
６.２ データメモリ ··················································································································································
データメモリの構造 ····························································································································
データの読出し ···································································································································
データメモリの設定 ····························································································································
データメモリのデータ消去 ··············································································································
データメモリのプリンタ出力 ············································································································

6 - 3
6 - 3
6 - 4
6 - 4
6 - 5
6 - 5

６.３ CF カード ·······················································································································································
データの読出し ···································································································································
ファイル名・ファイル形式の設定 ··································································································
ファイル削除 ········································································································································
設定ファイルの保存 ··························································································································
設定ファイルの読込み ·····················································································································
CF カードのフォーマット ···················································································································

6 - 6
6 - 6
6 - 7
6 - 9
6 - 9
6 -10
6 -10

６.４ ファイルのコピー ········································································································································ 6 -11
６.５ データメモリ・CF カードへの記録 ········································································································· 6 -12

第７章 インターフェースの設定
７.１ インターフェースの設定概要 ················································································································· 7 - 2
７.２ RS-232C の設定 ········································································································································ 7 - 3
通信条件の設定 ································································································································· 7 - 3
７.３ データの出力 ··············································································································································· 7 - 4
機器との接続 ······································································································································· 7 - 4
データ出力先の設定と出力方法 ································································································· 7 - 4
７.４ データ出力形式の設定 ··························································································································· 7 - 5
７.５ 外部表示器の設定 ···································································································································· 7 - 6
７.６ プリンタの設定の注意 ······························································································································ 7 - 6
７.７ リモート測定について ······························································································································· 7 - 7
構成 ························································································································································· 7 - 7
機能 ························································································································································· 7 - 7

第８章 その他の設定
８.１ その他の設定概要 ···································································································································· 8 - 2
８.２ オートパワーオフの設定 ························································································································· 8 - 3
８.３ バージョン情報 ············································································································································ 8 - 4
８.４ 日付・時刻の設定 ······································································································································ 8 - 4
８.５ 日本語/英語の切替え ····························································································································· 8 - 5
８.６ メンテナンス ·················································································································································
LCD バックライトの設定 ··················································································································
LCD バックライトの明るさ ···············································································································
コントラストの設定 ·····························································································································
ブザー音量の設定 ·····························································································································
バージョンアップ ·································································································································

8 - 6
8 - 6
8 - 7
8 - 7
8 - 8
8 - 8

８.７ 工場出荷時設定 ········································································································································· 8 -10
出荷時設定の実行 ···························································································································· 8 -10出荷時設定の一覧 ···························································································································· 8 -11

第９章 測定点数拡張
９.１ 測定点数拡張の概要 ······························································································································· 9 - 2
測定点数拡張機器 ···························································································································· 9 - 2
９.２ スイッチボックス ··········································································································································
CSW-5B の接続 ·································································································································
データメモリ ··········································································································································
測定モードの設定 ······························································································································
チャンネルの設定 ······························································································································
ボックスナンバの設定 ······················································································································
モニタ表示 ············································································································································
モニタの種類の選択 ·························································································································
初期値の処理 ······································································································································
プログラム設定 ···································································································································
測定値の記録 ······································································································································
各種チェック ·········································································································································

9 - 3
9 - 3
9 - 3
9 - 4
9 - 4
9 - 4
9 - 5
9 - 5
9 - 6
9 - 6
9 - 7
9 - 7

９.３ 傾斜計アダプタ ···········································································································································
IA-32 の接続 ·······································································································································
データメモリ ··········································································································································
測定モードの設定 ······························································································································
チャンネルの設定 ······························································································································
モニタ表示 ············································································································································
プログラム設定 ···································································································································
測定値の記録 ······································································································································
傾斜計モードでできないこと ··········································································································

9 - 8
9 - 8
9 - 8
9 - 9
9 - 9
9 - 9
9 -10
9 -10
9 -11

第１０章 ひずみの補正方法
１０.１ ひずみゲージの結線方法 ················································································································· 10 - 2
1 ゲージ 4 線式 ································································································································· 10 - 2
1 ゲージ法 2 線式(1/4 ブリッジ) ································································································· 10 - 2
1 ゲージ法 3 線式(1/4 ブリッジ) ································································································· 10 - 2
2 ゲージ法(1/2 ブリッジ) ··············································································································· 10 - 2
4 ゲージ法（フルブリッジ） ············································································································· 10 - 2
１０.２ センサケーブルの延長による感度低下 ······················································································ 10 - 3定電圧方式による測定 ·················································································································· 10 - 3定電流方式による測定 ·················································································································· 10 - 5
１０.３ ひずみの完全な補正方法（Comet） ······························································································ 10 - 7
1 ゲージ法 3 線式での補正方法 ······························································································· 10 - 7
Comet NON ········································································································································ 10 - 8
Comet A ··············································································································································· 10 - 8
Comet B (1 ゲージ法 3 線式のみ) ···························································································· 10 - 8
１０.４ リード線抵抗の求め方 ························································································································ 10 - 9
１０.５ 1 ゲージ 4 線式での補正方法 ········································································································· 10 -10

第１１章 仕様
１１.１ 仕様 ···························································································································································· 11 - 2
１１.２ 標準付属品 ············································································································································· 11 - 6

１１.３ オプション ················································································································································· 11 - 7
１１.４ 外観図 ······················································································································································· 11 – 8

第１２章 エラーメッセージ
１２.１ エラーメッセージの解説と対処方法 ······························································································ 12 - 2✎memo１章

第概要
１.１ 概要 ······················································································

1 - 2

１.２ 特長 ······················································································

1 - 2

１.３ 各部の説明 ·······································································

1 - 3

１．１ 概要

１．１ 概要
片手にすっきり収まるコンパクトな形状のハンドヘルドタイプの測定器です。防滴構造を採用していますので、屋外でも安心して使用できます。センサを接続する端子部は、リード線、バナ
ナプラグとも簡単に接続可能なワンタッチタイプ（特許）を採用し、スピーディに測定ができます。最大 20 チャンネルのセンサモード、係数、イニシャル値の設定と測定データの記録が可能で、複数の現場を移動しながらデータを収集する場合などでもデータ整理が容易です。また、スイッ
チボックスを使用することにより、5 点の自動測定が可能です。インターバルタイマ、データメ
モリ、さらには CF カードによる記録や、パソコンと接続してコントロールやデータ転送も可能です。抵抗および絶縁抵抗のチェック機能により、ひずみゲージや変換器などのチェックにも使用できます。

１．２ 特長・ひずみ、直流電圧、熱電対、白金測温抵抗体、抵抗、絶縁抵抗が測定可能
・絶縁抵抗測定もできるのでセンサのチェックも可能

・1G4W 対応（1 ゲージ 4 線式ひずみ測定法）
・TEDS 対応
・ばら線の接続もワンタッチ
・インターバルタイマによる自動測定・低消費電力・電源は単 3 形電池 4 本で、現場での交換が容易
・スイッチボックスとの組合せで多点測定
・2 軸傾斜計用アダプタとの組合せ可能

1-2

１．３

各部の説明

１．３ 各部の説明❏ 前面、側面
①

②

⑩
⑫
⑪

③

⑬
⑭

⑤
⑥

④

⑦

⑧
⑨

① 入力端子② メイン LCD
③ サブ LCD④ POWER キー⑤ ファンクションキー⑥ カーソルキー⑦ テンキー⑧ ENT キー
⑨ ESC キー
⑩ I/F コネクタキャップ⑪ DC コネクタキャップ⑫ USB コネクタ
⑬ RS-232C コネクタ
⑭ DC 入力コネクタ

ひずみゲージや各種センサを接続します。
測定値のモニタ表示や各種設定をします。
タイマの動作やメモリの状況等を表示します。
主電源スイッチです。
ファンクションを切替えます。
カーソルを移動します。
数値の入力、メニュー項目の選択に使用します。
設定値の確定、タイマのスタート等に使用します。
設定の取消し、各メニューからのエスケープ等に使用します。
インターフェースコネクタの保護キャップです。
DC 入力コネクタの保護キャップです。
USB 接続用コネクタです。
RS-232C 接続用コネクタです。
AC アダプタを接続します。

1-3

１．３ 各部の説明❏ 上面、底面、背面
⑥

①
⑦
⑧
②

⑨

③

⑤

④

⑩

① 結線シール② 入力コネクタ③ CF カード スロット
④ イジェクトボタン⑤ CF カバー⑥ NDIS コネクタキャップ⑦ 固定用ネジ⑧ 滑り止めラバー⑨ 製造番号⑩ バッテリハッチ

1-4

ひずみゲージの結線方法を示します。
各種センサを接続する NDIS コネクタです。
CF カード TYPEⅠ規格のメモリカード挿入口です。
CF カードを取出すダブルプッシュタイプのボタンです。
CF カードスロットを保護するカバーです。
NDIS コネクタを保護するキャップです。
本体を固定する場合に使用します。(有効ネジ深さ 5mm)
滑止め用のゴムシートです。
製造番号です。
バッテリ交換時に外します。

２章

第

準備
２.１ 使用上の注意 ··································································

2 - 2

２.２ 電源について ····································································

2 - 3

２.３ 現場測定での注意 ·························································

2 - 6

２.４ 画面表示の概要 ······························································

2 - 7

２.５ 操作の概要 ·······································································

2 - 8

２．１ 使用上の注意

２．１ 使用上の注意
本器を使用する上で、注意していただきたい項目です。
注意 本器はコネクタキャップをした状態で IP-54（いかなる方向からの水の飛沫によっても有害な影響を受けない）相当となっています。直接水中に没するようなことは避けてください。また、コネクタキャップ、および CF カバーをはずした状態では防水になりませんので、ご注意ください。
注意 大きな振動や連続して振動するところで使用したり、運搬する時に落としたり、強い衝撃を加えると故障の原因になります。
注意 運送する場合はご納入時の梱包材を使用するなど、必要に応じて振動、衝撃から守る対策を施してください。

注意 本器の上に重いものを載せないでください。
注意 筐体が汚れた時は、柔らかい布などに薄めた中性洗剤をつけて軽く拭き取り、よく乾拭きしてください。シンナーなどの強い溶剤は筐体が溶けたり、変色することがありますので使用しないでください。
注意 CF カードを抜き取るときに使用するイジェクトボタンは、飛び出した状態にあると外力により壊れやすくなります。通常は中に押し込んだ状態でご使用ください。スロットへは、CF カ
ード以外のものを入れないでください。

2-2

注記

液晶表示器は多数の画素から構成されており、常時点灯する画素、点灯しない画素が存
在することがあります。これをドット抜け・画素欠けと呼びますが、故障ではなく、液晶表示器の性質によるものですので、予めご了承ください。また、ドット抜け・画素欠けは初期不良や修理、交換の対象とはなりません。重ねてご了承ください。

注記

CF カードの抜き挿しは、5 秒以上の間隔を開けて行ってください。CF カードの認識が正常にできない場合があります。また、データ書込み中は、CF カードを抜いたり、電源を OFF にしないでください。

２．２

電源について２．２ 電源について
単 3 形の電池 4 本、または別売の AC アダプタにて駆動することができます。

❏ 電池の入れ方
電池は単 3 形のアルカリ乾電池や、充電式の電池を使用することができます。本器に充電する
機能はありませんので、充電済みの電池をご使用ください。
①コインなどでネジを半時計方向に回して、バッテリハッチをはずします。

ゆるむ②極性に注意しながら、電池を 4 本入れます。
③はずした時と逆の要領で、バッテリハッチ
を元に戻します。

注意 電池の破裂、液漏れの恐れがありますので、○＋、－の極性を正しく入れてください。
○種類の異なる電池、新旧の電池を混合しないでください。
○長期間使用しない場合は、本体から電池を取出してください。

2-3

２．２ 電源について❏ 電池での使用時間
電池での使用時間は、周囲温度等により異なります。下表にて、電池が新品（アルカリ乾電池）の場合の標準的な連続使用時間を示します。
○連続使用時間条件

アルカリ乾電池

TC-32K 単体

10 時間

TC-32K ＋ CSW-5A・B

6 時間※オートパワーオフ機能を使用しない場合※LCD バックライト OFF※モニタ表示 ON※CF カード挿入※周囲温度 23℃

○スリープインターバル時の使用時間
周囲温度 23℃
TC-32K 単体＋CSW-5A・B

TC-32K 単体＋CSW-5A・B

1分

60 時間（2.5 日）

43 時間（1.8 日）

42 時間（1.75 日）

30 時間（1.2 日）

10 分

580 時間（24 日）

428 時間（17 日）

400 時間（16 日）

300 時間（12 日）

1 時間

2800 時間（116 日）

2400 時間（100 日）

1960 時間（81 日）

1680 時間（70 日）

3 時間以上7200 時間（300 日）

6000 時間（250 日）

5000 時間（208 日）

3500 時間（145 日）※アルカリ乾電池使用

2-4

周囲温度 0℃

インターバル時間

２．２

電源について❏ AC アダプタの接続
AC アダプタ｢CR-1867｣は別売です。
コネクタキャップをはずし、AC アダプタ
「CR-1867」を本体の DC 入力コネクタに接続します。次に、AC アダプタ本体をAC100～240V コンセントに差込みます。
電池を使用している場合は、AC アダプタ
からの電源供給を優先します。

CR-1867

AC100～240V
コンセントへ

注意 ○｢CR-1867｣以外の AC アダプタは使用しないでください。
○AC アダプタは、本体に接続してから AC100～240V コンセントに差込んでください。

❏ 電源の ON/OFF○電源 ON【POWER】キーを 2 秒以上押します。メイン LCD に起動画面を表示し、次にモニタ画面を表示します。

○電源 OFF【POWER】キーを 2 秒以上押します。

注記

極端に早い電源の ON/OFF 操作の繰返しは、本器に負担をかけます。5 秒以上の間隔をおいて行ってください。

2-5

２．３ 現場測定での注意

２．３ 現場測定での注意
スイッチボックスと組合わせ、据置きの状態で長期の測定を行う場合は、必ずアースを接続してください。また、落雷などの可能性がある場合においても、確実にアース処理を施してください。

❏ アースの接続
入力端子の E にアース線を接続してください。アース
線はできるだけ太いものを使用し、かつ最短で接地してください。接地は、雷や重機などからのノイズ対策にも有効です。

危険 雷が接近してきた時、また落雷の可能性がある時は、アースや電源などの配線工事、測定器の操作などを中断して、機器から離れてください。これらの作業中に落雷した場
合、感電、やけど、時には生命の危険が生じる可能性があります。

❏ 落雷対策スイッチボックスとセンサ間のケーブルを延長している現場では、たとえ直撃雷でなくとも、落雷の強力な誘導作用で発生したパルスにより、センサやスイッチボックス、測定器の破壊にいたるなど深刻な影響を受ける場合がありますので、適切な対策が不可欠となります。
○ケーブル敷設センサケーブルはゴム被覆、シールド外被付きの変換器用ケーブルなどを使用します。設置の際には、大地との誘導電位差を少なくするために、空中配線を避け、地面に這わせるようにします。鉄塔への設置などケーブルを地面に這わせることができない場合は、次のセンサ
用避雷器を併用すると有効です。
○センサ用避雷器（アレスタ）
センサ用避雷器｢NZ-6B｣は変換器に対応した 6 線式で、ひずみゲージにも適用が可能です。
この避雷器はなるべくセンサに近い部分に挿入してください。なお、「TC-32K」の入力端子にセンサ用簡易型避雷器を内蔵していますが、万全を期するためにもセンサ用避雷器を併用することを推奨します。

2-6

２．４

画面表示の概要

２．４ 画面表示の概要❏ 起動画面について電源を投入してまもなく、左の起動画面を表示します。この画面の次に下記のモニタ画面を表示します。

❏ 画面構成について
起動後に表示するモニタ画面を中心に、機能や測定動作などの設定画面へ移行します。下図に、画面構成の概略を示します。
モニタ画面で数値や状態確認を行い、設定変更時には「メニュー」を選択して、各種の設定画面へ移行するというのが基本的な使い方になります。

モニタ画面○測定値の表示○メインの画面

初期値の処理○イニシャルイン○モニタチャンネルのイニシャルイン○初期値の書換え

チャンネル設定○測定チャンネル選択○測定モードの選択

メニュー○プログラム設定○メジャー/ダイレクトの切替え○自動測定の設定○各種チェック○測定に関する設定
○記録
他

2-7

２．５ 操作の概要

２．５ 操作の概要
モニタ画面を中心に、3 つのファンクションキーから各種のモードへ移行します。このファン
クションキーとモードの体系、および操作に使用するキーとその機能について説明します。

❏ 操作体系
TC-32K の操作体系を示します。
モニタ画面
初期値の処理イニシャルインモニタチャンネルのイニシャルイン
初期値の書換えチャンネル設定
測定モードの切替え
CSW-5B の設定※マルチチャンネル設定時のみ
モニタの種類の選択
メニュー
プログラム設定
センサモードの設定
係数、表示桁、単位の設定
係数
表示桁
単位
Cap/Ro の設定
係数設定チェック
熱電対基準接点の設定
TEDS センサの読込み測定モードの切替え
メジャー/ダイレクトの切替え
自動測定の設定
インターバル測定の設定
インターバルの開始･停止
各種チェック
絶縁チェック
抵抗チェック
ばらつきチェック
リード線抵抗チェック
ブリッジ出力チェック
係数設定チェック
測定に関する設定
シンプルメジャーの設定
Comet の設定
測定環境の商用電源周波数
イニシャル値測定の設定
バーンアウトチェックの設定

2-8

メニュー
記録
データメモリ
データメモリの読出し
データメモリの設定
データ番号の設定
リングバッファの設定
データメモリのデータ消去
データメモリのプリンタ出力
CF カード
データの読出しファイル名・ファイル形式の設定
ファイル削除設定ファイルの保存
設定ファイルの読込み
CF カードのフォーマット
ファイルのコピー
データメモリ・CF カードへの記録
インターフェースの設定
RS-232C の設定
ボーレート設定
データビット設定
パリティ設定ストップビット設定
フロー制御設定
タイムアウト設定
データ出力先の設定
データ出力形式の設定
データフォーマット
ヘッダの有無タイムデータの有無
外部表示器の設定
ボーレート設定
パリティ設定その他の設定オートパワーオフの設定
バージョン情報
日付・時刻の設定
日本語/英語の切替え
メンテナンスLCD バックライトの設定
LCD バックライトの明るさ
コントラストの設定
音量の設定
バージョンアップ
工場出荷時設定

２．５

操作の概要

❏ キースイッチ
メニューへの移行を始めとして、画面の切替えや数値入力、測定の開始やチェックなど、前面に配置しているキースイッチを使用して、操作を進めていきます。
本文中における手順の説明には、「【F3】キーを押します」のように操作するキースイッチを記述
しています。各キーの主な機能は、以下の通りです。

操作キー

主な機能

F1

初期値の処理へ移行

F2

チャンネル設定へ移行、ページ切替え

F3

メニューへ移行、ページ切替え

▴▾◂▸
0～9
ENT／STARTESC／KEY-LOCK

カーソルの移動、モニタチャンネルの選択、数値の増減
数値入力、メニュー項目選択
設定の確定、スキャニング測定の開始設定の取消し、メニュー階層の浮上、キーロック実行、解除❏ キーロック
不用意な操作や設定の変更を防止する機能として、キーロックがあります。ただし、【ENT/START】キーと【POWER】キーは、キーロック中でも操作可能です。
○キーロック有効【ESC／KEY-LOCK】キーを 3 秒以上押します。ブザー音とともに、キーロックが有効になります。また、画面にはキーロック中を示すマークを表示します。

○キーロック解除【ESC／KEY-LOCK】キーを 3 秒以上押します。ブザー音とともに、キーロックを解除
します。

2-9

２．５ 操作の概要

✎memo

2-10

３章

第

センサの接続
３.１ センサの接続 ····································································

3 – 2

３．１ センサの接続

３．１ センサの接続
センサの接続には、ワンタッチ方式の端子台と NDIS コネクタの 2 通りがあります。ワンタッチ方式の端子台は、センサケーブルがバラ線の状態であるものに適用し、細い線から圧着端子付きのもの、バナナプラグのタイプまで幅広く対応することが可能です。

❏ 入力端子台の使用方法
入力端子台の各部の名称、および適用するリード線や線端形状と端子台のレバー操作について、説明します。
○各部名称
板バネ

挿入口

レバー
端子台

○結線方法
結線対象線

レバー操作

極端に細い
リード線

説明

レバーを手前に倒します。端子台と板バネの間にリード線の導体部
分、もしくは圧着端子の矢型部分を挿入し、レバーを戻します。

矢型圧着端子
倒す

倒す

リード線

戻す

レバーを奥側に倒します。端子挿入口にリード線を通し、レバーを戻します（極端に細いリード線の場
合、切れることがあります）。

倒す

バナナプラグ

注記

3-2

レバーを奥側に倒し、端子挿入口にバナナプラグを差込みます（レバーを戻す必要はありません）。

レバー操作が堅くなった場合、アルコール等で粉塵などを洗浄した後、シリコングリス、または自動車用のモリブデン配合のグリス等でグリスアップしてください。

３．１

センサの接続❏ センサ別結線方法
測定対象となるセンサの種類により、結線方法が異なります。以下にセンサ別の結線方法、および選択するセンサモードを示します。
○結線表
測定対象

結線図

B-C 間をリード線でショート

1 ゲージ法 2 線式

センサモード

1G3W 120Ω
1G3W 240Ω
1G3W 350Ω

1 ゲージ法 3 線式

1G3W 120Ω
1G3W 240Ω
1G3W 350Ω

1 ゲージ 4 線式

1G4W 120Ω
1G4W 240Ω
1G4W 350Ω

モジュラーコネクタ

1 ゲージ 4 線式
変換アダプタ（別売）

2 ゲージ法4 ゲージ法1 ゲージ 4 線式
変換アダプタ

1G4W 120Ω
1G4W 240Ω
1G4W 350Ω

2GAGE

4GAGE
4G C350Ω
4G 0-2V

3-3

３．１ センサの接続

測定対象

結線図

4GAGE
4G C350Ω
4G 0-2V

変換器
NDIS コネクタ付

変換器
先バラ線

センサモード赤 緑

4GAGE
4G C350Ω
4G 0-2V

黒白※弊社製変換器の場合

熱電対
温度測定

直流電圧測定白金測温抵抗体

注記＋－

熱電対 T、K、J、B
S、R、E、N

直流電圧 300mV
直流電圧 30V
直流電 AUTO（～30V）

Pt100 3W

キャリブレータなど、入力コネクタの F、G をリモートセンス用に結線してあるものは使用できません。

注意 変換器のリード線の色は、メーカーにより異なります。弊社製以外の変換器を使用する場
合は、変換器の取扱説明書をよくご確認ください。

3-4

第

４章

モニタ表示と測定
４.１ モニタ画面の概要 ··············································

4 - 2

４.２ モニタ表示 ···························································

4 - 2

４.３ 初期値の処理 ·····················································

4 - 5

４.４ チャンネル設定 ···················································

4 – 8

４.５ 測定値の記録 ·····················································

4 -10

４.６ サブ LCD ······························································

4 -11

４．１ モニタ画面の概要

４．１ モニタ画面の概要
電源投入後、ディスプレイに表示する画面がモニタ画面です。この画面では、測定値の表示（モニタ）を始め、時間やステータスなどを表示します。さらに、キースイッチを使用
して、さまざまな機能の詳細設定画面へ移行します。
RS-232C にて通信中
USB にて通信中
CF カードに記録
データメモリに記録

シンプルメジャー
キーロック中

日付、時刻

バッテリ残量表示（バッテリ消 耗 に伴 い、点灯領域が減少します）

モニタスペース（モニタ値を表示します）

ファンクションキー対応【メニュー】（メニュー画面へ移行します）【初期値の処理】（初期値の処理画面に移行します）【チャンネル設定】（モニタするチャンネルを設定します）

４．２ モニタ表示
モニタ表示とは、測定している値を画面へ常時表示する機能です。あらかじめ設定したモニタ表示の方法に応じて、ディスプレイに表示します。
表示方法には、数値モニタ、波形モニタ、モニタ OFF 表示の 3 種類があります。数値モ
ニタは 1 チャンネル、専用スイッチボックスと接続して 5 チャンネル(マルチチャンネルモー
ド選択時)の表示タイプから選択できます。以下に、モニタ画面の例を示し、画面内の各項目について説明します。

❏ 数値モニタ
選択したチャンネルの測定値を数値で表示します。画面に表示できるチャンネル数は、最
大 5 チャンネル分（マルチチャンネルモード選択時）です。
○1 チャンネル表示の例（シングルチャンネルモード）
D：ダイレクト
M：メジャー
m：メジャー（Comet）
J：ジャンプ

測定対象（シンプルメジャー設定時）
センサモード
測定値（モニタ値）
単位

4-2

４．２

モニタ表示

○5 チャンネル表示の例（マルチチャンネルモード）
ボックスナンバ

CSW-5B のチャンネル

モニタ中を示すマーク❏ 波形モニタ
測定値の時間変化を、折れ線グラフで表示します。波形モニタでモニタできるチャンネル
は 1 チャンネルです。

モニタチャンネル

測定値（モニタ値）❏ 表示値の意味
入力端子にセンサを接続しており、センサや測定系に異常がなければ何らかの数値を表示します。しかし、センサを接続していない、接続しているが途中で断線している、入力値が仕様の測定範囲を超えている場合などにおいては、測定値に代わって以下のように異常を表示します。異常がある場合の表示と原因は次の通りです。
○「＊＊＊＊＊＊＊＊」
オープンデータ
入力が接続していないか断線しています。
○「＋＊＊＊＊＊＊＊＊」＋オーバデータ
入力値が測定範囲を＋側に超えています。
○「－＊＊＊＊＊＊＊＊」－オーバデータ
入力値が測定範囲を－側に超えています。
○「＊＊＊＊I＊＊＊」
初期値オーバデータ
初期値（初期不平衡値）が初期値記憶範囲を超えています。
○「！！！！！！！！」
表示桁オーバ
測定値に対して表示する桁（小数点以上）が足りません。
○「――JUMP――」
ジャンプ
センサモードが JUMP の場合に表示します。

4-3

４．２ モニタ表示❏ モニタの種類の選択
モニタ値の表示形式を数値、波形、または OFF から選択できます。モニタ OFF にすると、センサへの電源供給を遮断し、ディスプレイにはモニタ値を表示しません（測定をしません）。
①モニタ画面から、【F2】キーを押して、「チ
ャンネル設定」に移行します。

②チャンネル設定画面を表示します。
【F3】キ ーを押 して 、「 測 定モードの切替
え」に移行します。

③さらに【F3】キーを押して、「 モニタの種類の選択」に移行します。

◀ ▶

④【 】【 】キーでカーソル(反転表示)を移動して、【ENT】キーを押してモニタの種類を選択します。

❏ 波形モニタの設定
波形モニタを選択した場合は波形モニタの設定ができます。波形モニタでモニタできるチ
ャンネルは 1 チャンネルです。
○波形モニタの設定
モニタ画面から、【F2】キーを押して、「モニタの設定」に移行すると、左の画面を表示します。各項目の設定値を【 】【 】
キーで切替えることができます。

◀ ▶

モニタチャンネル：0～19
Y 軸設定：表示桁の設定により可変
Y 軸ゼロ点設定：0、1、2、3、4
T 軸設定：10～100 秒

4-4

４．３

初期値の処理４．３ 初期値の処理
イニシャルインとは、現在の測定値を記憶する機能です。これはひずみゲージ、ひずみゲ
ージ式変換器などの初期不平衡値（初期値またはイニシャル値）を差引いて測定・表示する場合に使用します。初期不平衡値とは、外力を受けていないにもかかわらず、ゲージ抵
抗値の微小なズレによって発生する信号のことです。また 変位計等では、設置した位置を基準にして変位を測定する場合、基準の位置でイニシャルインを実行します。これにより、基準からの相対変位量を得ることができます。
測定値を表示する方法には初期値を差引いて表示するメジャー値（モニタ表示上「M」
または「m」が付きます）、初期値を差引かないダイレクト値（モニタ表示上「D」が付きます）があります。ひずみ測定と同様に、電圧測定の場合も初期値を差 引いて表示することが可能です。ただし、温度（熱電対・白金測温抵抗体）の場合は、初期値を差 引いて表示することはできません。

❏ イニシャルイン
測定中のチャンネルに対して、初期値を取ります。
① モ ニ タ 画 面 か ら 、 【F1】キ ー を 押 し て 、「初期値の処理」に移行します。

②【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【1】を押して、「イニシャルイン」に移行します。

◀

③確認のメッセージを表示します。【 】【 】キーで「はい」にカーソルを移動して、【ENT】キーを押します。

▶

④初期値を取って、測定値を表示します。
（「D」が「M」に変わります。）

注記

イニシャルインの許可で、【禁止】を選択している場合は、初期値を取ることはできません。

4-5

４．３ 初期値の処理❏ モニタチャンネルのイニシャルイン
マルチチャンネル時にモニタ測定中のチャンネルに対して、初期値を取ります。

① モ ニ タ 画 面 か ら 、 【F1】キ ー を 押 し て 、「初期値の処理」に移行します。
注記 イニシャルインの許可で、【禁止】を選択している場合、初期値を取ることはできません。

②【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【2】を押して、「モニタチャンネルのイニ
シャルイン」に移行します。
注記 マルチチャンネル時のみ動作します。
シングルチャンネルモードや、モニタ
OFF の場合、モニタチャンネルのイニシャルで初期値を取ることはできません。

◀

③確認のメッセージを表示します。【 】【 】キーで「はい」にカーソル（反転表示）を移動して、【ENT】キーを押します。

▶

④初期値を取って、測定値を表示します。

（「D」が「M」に変わります。）

4-6

４．３

初期値の処理❏ 初期値の書換え
センサが有している初期不平衡値とは別に、任意に初期値を変更することができます。
① モ ニ タ 画 面 か ら 、 【F1】キ ー を 押 し て 、「初期値の処理」に移行します。
②【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【3】を押して、「初期値の書換え」に移行します。

③【▲】【▼】キーで、書換えるチャンネルの初期値にカーソル（反転表示）を移動して【ENT】キー押します。

注記

イニシャルインを実行している場合は、初期値にその値を表示します。

④テンキーを用いて直接入力します。符号の切替えには【F1】キー、一旦クリアする
場合は、【F3】キーを使用します。入力後、【ENT】キーを押します。

⑤初期値が入力した値に書換わります。
【ESC】キーでモニタ画面に戻ります。

4-7

４．４ チャンネル設定

４．４ チャンネル設定
本器は、測定点数が１点の測定器です。しかし、点在しているセンサの測定データを収集する場合は、測定対象によってセンサが異なるとともに、同じひずみゲージやひずみゲー
ジ式変換器であっても、初期値が異なります。本器では、チャンネルという設定を通じて、センサモードの情報や物理量直読のためのパラメータ、さらに初期値のデータを 20 通り記
憶することができます。
チャンネルの使用に当たっては、接続するセンサやスイッチボックスなどにより、測定モ
ードを切替える必要があります。マルチチャンネルモード、傾斜計 1 軸モード、および傾
斜計 2 軸モードについては、｢第９章 測定点数拡張｣を参照してください。

❏ 測定モードについて測定モードには、下記の４種類があります。測定形態に合わせて、モードを選択してください。
○シングルチャンネルモード
端子台や NDIS コネクタを使用して、本器に接続するセンサが 1 つの場合に選択します。
チャンネル 00～19 まで、20 チャンネル分のパラメータと初期値を設定・記憶することができます。
○マルチチャンネルモード
オプションのスイッチボックス、「CSW-5A」、「CSW-5A-05」、「CSW-5B」、「CSW-5B-05」
を使用する場合に選択します。
測定点数は 5 点に拡張され、この 5 点のモニタ表示や自動測定が可能です。
○傾斜計 1 軸モード
弊社製の挿入型傾斜計を測定する場合に選択します。
○傾斜計 2 軸モード
オプションの 2 軸挿入型傾斜計アダプタ「IA-32」を介して、弊社製の挿入型傾斜計を測定する場合に選択します。同時に 2 軸分の測定データを表示、記録することが可能です。

❏ 測定モードの切替え①モニタ画面から、【F2】キーを押して、「チ
ャンネル設定」に移行します。

4-8

４．４

チャンネル設定② 現 在 の 測 定 モ ー ド に対 応 し た チ ャ ン ネ
ル設定画面を表示します。【F3】キーを押して、「測定モードの切替え」に移行します。

③ 【▲】【▼】キ ーで カ ーソ ル (反 転表 示 )を移動して、モードを選択します。
④【ENT】キーを押します。設定したモードのチャンネル設定画面を表示します。

❏ シングルチャンネルモードのチャンネル設定
モニタ表示するチャンネルを変更します。
①モニタ画面から、【F2】キーを押して、「チ
ャンネル設定」に移行します。
② 現 在 の モ ニ タ 中 の チャ ン ネ ル を 表 示 します。

③ カ ー ソ ル 位 置 の 数 字 を 【▲】【▼】キーで増減、もしくはテンキーで直接入力し、【ENT】キーを押します。

④ 設 定 し た チ ャ ン ネ ルを モ ニ タ 表 示 し ます。

4-9

４．５ 測定値の記録

４．５ 測定値の記録
モニタ表示だけでは、測定値を記録しません。測定値を記録する方法には、前面パネルの【ENT/START】キーを押して記録する手動測定、自動で記録を行う自動測定（インターバ
ル測定）があります。

❏ 手動測定モニタ表示している測定値を、データメモリや CF カードに記録します。また、外部機器を接続している場合は、データを出力します。
①データメモリへの記録、もしくは CF カードへの記録が「On」の状態であることを確認してください。詳細については、｢第６章 記録の設定｣を参照してください。

②【ENT/START】キーを押します。
③測定値を設定した媒体に記録します。

ⒶⒷ○データ番号の見方
デー タメモリへ の記録を 「On」にすると左のように、モニタ画面の右上にデータ番号を表示します。
Ⓐ：次に記録するデータのデータ番号Ⓑ：記録可能なデータ数（測定回数）
データメモリの空き容量がなくなると「DM：Full」と表示します。
注記

4-10

記録可能なデータ数は、測 定モードにより異なります。

４．６

サブ LCD❏ 自動測定モニタ表示している測定値を、設定した時間間隔、あるいは設定した時刻にて、自動でデータメモリや CF カードに記録します。インターバル測定の設定や、インターバル測定の開始・停止などの詳細については｢５．８ 自動測定｣を参照してください。

４．６ サブ LCD
バッテリ電圧低下、自動測定中、データメモリの空き容量なし、AC アダプタ使用中の場合に点灯します。

データ メモ リの空 き容 量 がな い場 合に点灯します。

AC アダプタ接続時に点灯します。

バッテリ電圧低下時に点灯します。
自動測定中、1 秒毎に順次点灯します。

注記○自動測定中を示すインジケータは、電源 OFF（スリープ ON にて自動測定開始）の状
態でも、点灯します。
○バッテリ電圧低下を表示した場合は、速やかに新品の電池と交換してください。

4-11

４．６ サブ LCD

✎memo

4-12

第

５章

測定の設定
５.１ 測定の設定概要 ················································

5 - 2

５.２ センサモードの設定 ···········································

5 - 3

５.３ 係数・表示桁・単位 ············································

5 - 6

５.４ 熱電対基準接点の設定 ···································

5 -11

５.５ TEDS センサ ·······················································

5 -12

５.６ 測定モードの切替え ··········································

5 -13

５.７ メジャーとダイレクトの切替え ··························

5 -14

５.８ 自動測定 ·····························································

5 -15

５.９ 各種チェック ························································

5 -22

５.１０ 測定補助設定 ················································

5 -24

５．１ 測定の設定概要

５．１ 測定の設定概要
センサに対する詳細な設定を始め、自動測定や各種のチェックなど、測定に関する操作、設定について説明します。これら各種設定画面の構成を以下に示します。

【F3】キーを押してメニューへ移行します。【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーを押して、各設定画面に移行します。さらに、カーソルの移動、もしくはテンキーにて詳細設定項目を選択します。

5-2

５．２

センサモードの設定

５．２ センサモードの設定
接続しているセンサに合わせて、センサモードや係数・単位の設定を行います。また、
TEDS センサの読込み、測定モードの切替え等について説明します。

❏ センサモードについて
ひずみゲージ、ひずみゲージ式変換器を使用するためには、各入力機器に合ったセンサ
モードを設定する必要があります。
○センサの種類
接続するセンサの種類を設定します。以下に、設定可能なセンサの一覧を示します。結線法については「１０．１ ひずみゲージの結線方法について」を参照してください。
センサモード一覧
測定対象

ひずみ測定

熱電対温度
測定

電圧測定
白金測温抵抗体温度測定

センサモード

センサ詳細

備考

1G4W 120Ω

1 ゲージ 4 線式 120Ω

1G4W 240Ω

1 ゲージ 4 線式 240Ω

1G4W 350Ω

1 ゲージ 4 線式 350Ω

1G3W 120Ω

1 ゲージ法 3 線式 120Ω

1G3W 240Ω

1 ゲージ法 3 線式 240Ω

1G3W 350Ω

1 ゲージ法 3 線式 350Ω

2GAGE

2 ゲージ法120～1000Ω

4GAGE

4 ゲージ法120～1000Ω

4G C350Ω

4 ゲージ定電流法

4GAGE 0-2V

4 ゲージ法 0-2V
120～1000Ω

ブリッジ電圧DC2V 24ms（50Hz）

JUMP

測定しない

対象チャンネルの測定をしない

熱電対 T(CC)

熱電対 T

熱電対 K(CA)

熱電対 K

熱電対 J(IC)

熱電対 J

熱電対 B

熱電対 B

熱電対 S

熱電対 S

熱電対 R

熱電対 R

熱電対 E（CRC）

熱電対 E

熱電対 N

熱電対 N

DC 300mV

電圧測定 ±300mVDC 30V

電圧測定 ± 30 V

DC AUTO

電圧測定 ± 30 V ※1

Pt100 3W

白金温度測定 3 線式

ブリッジ電圧DC1V 44ms（50Hz）

350Ω

リニアライズ
デジタル演算JIS C1602-1995

入力インピーダンス
V 1/1
500MΩ以上
V 1/100
1MΩ以上
リニアライズ デジタル演算
JIS C1604-1997 Pt100※1：本体からの 1 チャンネル測定のみ

5-3

５．２ センサモードの設定❏ センサモードの設定①メニュー画面から、【▲】【▼】キーでカー
ソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【1】を押して、「プロ
グラム設定」に移行します。

②【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【1】を押して、「センサモード設定」に移行します。

③ 【▲】【▼】キーで設 定 す る チ ャ ン ネ ル の
セ ン サ モー ド にカ ーソル を 移 動 して、【ENT】キーを押します。

④センサモードの一覧を表示します。【▲】【▼】【 】【 】キーで設定するセンサモー
ドにカーソルを移動して、【ENT】キーを
押します。

◀ ▶

【F2 】【F3】キーで一覧表示のページを切
替えます。

注記

5-4

センサモードに 1ゲージ法 3線式を選択し、かつ Comet A、B を使用する設定の場合
は、初期不平衡値やリード線抵抗の両端電圧に対して、自動的に補正演算を 行います。よって、これらのチャンネルはメジャー値に設定し、モニタ表示上は「m」が付きます。Comet の詳細については、「１０．３ ひずみの完全な補正方法（Comet）」を参照してください。

５．２

センサモードの設定❏ グループ設定について
センサモードの設定を始め、後述の各設定の操作方法において、複数のチャンネルの設定を一括して行うことができます。これをグループ設定と呼び、あるチャンネルからあるチ
ャンネルまでを 1 つのグループとして扱います。
①各種の詳細設定画面から【F3】キーを押して、各項目のグループ設定に移行します。

②テンキー、もしくは【▲】【▼】キーを使用して、「From」と「To」に設定チャンネ
ルの範囲（グループ）を指定します。続いて、【 】【 】キーで設定値にカーソルを移動して、【ENT】キーを押します。

◀ ▶

③設定値の選択、もしくは数値の入力を行った後、【ENT】キーを押します。

④指定した範囲のチャンネルに対して、一括設定します。

5-5

５．３ 係数・表示桁・単位

５．３ 係数・表示桁・単位❏ パラメータについて
ひずみゲージ、ひずみゲージ式変換器を用いて、応力、荷重、変位や圧力等を測定し、物理量で測定値を表示するためには、以下に示すパラメータ（係数、単位、表示桁）の設定が必要です。
設定可能な係数の設定範囲○係数
測定値を物理量で表示するため、生データに乗ずる値です。設定可
能な範囲は右の通りです。
○単位
測定値を表示する場合や、記録するときにデータへ付加します。設
定可能な単位は右の通りです。

範囲±（0.0001 ～ 99999）

係数

設定可能な表示単位の一覧
με
μ
N
kN
MN
gf
kgf
tf
kgm
Nm

○表示桁
小数点以下 5 桁まで表示可能です。設定可能な表示桁は右の通り
です。

℃
˚F
mm
cm
m
hPa
kPa
MPa
N/mm ２
Kg/mm

ｋｇ/ｃｍ
m/s２
G
ｒｐｍ
Ｈｚ
μV
mV
V
μA
mA

A
Ω
kΩ
MΩ
deg
Tor
％
ppm
スペース

###

設定可能な表示桁一覧（６種類）
表示桁###########．#####．#####．#####．#####．#####❏ パラメータの設定例○ゲージ率の補正(ゲージ率：KG＝2.13 の場合)
2.000/2.13＝0.939 となるので係数＝0.939を設定します。
○単純応力測定(ゲージ率:KG＝1.98 ヤング率:Ｅ＝2100000 kgf/cm ２ （スチール）の場合)
2.000/1.98×2100000×10 － ６ ＝2.121 となるので係数＝2.121を設定します。

5-6５．３

係数・表示桁・単位○荷重計の表示を物理量に変換する(定格容量：5 kN、定格出力：2 mV/V のロードセルの場合)
定格出力が mV/V で表現しているものは、1 mV/V＝2000×10 － ６ ひずみと計算します。2mV/V×2000＝4000×10 － ６ ひずみとします。
5 kN/4000×10 － ６ ひずみ＝0.00125 となるので係数＝1.250×10 － ３ 、単位＝kN、表示桁＝###.###と設定すれば、表示値は物理量になります。

○変位計の表示を物理量に変換する(定格容量：25mm、定格出力：6.25mV/V の変位計の場合)
6.25mV/V×2000＝12500×10 － ６ ひずみ とします。
25/12500×10 － ６ ひずみ＝0.002 となり
係数＝2.000×10 － ３ 、単位＝mm、表示桁＝###.###と設定すれば、表示値は物理量になります。

○電圧出力センサの表示を物理量に変換する(定格容量：10mm、電圧出力 0～5V の場合)
10/5000mV＝0.002 となり、
センサモード＝直流電圧 30V、係数＝2.000× 10 － ３ 、単位＝mm、表示桁＝##.####と設定すれば、表示値は物理量になります。

○シャント抵抗の電圧降下を測定し、電流を表示する(測定電圧：300mV 以下、シャント抵抗 10Ωの場合)
1/10Ω＝0.1 となりセンサモード＝直流電圧 300mV、係数＝1.000× 10 － １ 、
単位＝mA、表示桁＝##.####と設定すれば、表示値は電流となります。

❏ 係数・表示桁・単位の設定①プログラム設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【2】を押して、「係数・表示桁・単位の設定」に移行します。

5-7

５．３ 係数・表示桁・単位② チ ャ ン ネ ル ご と の 設定 を 表 示 し ま す 。
【▲】【▼】【 】【 】キーで変更する項目にカーソルを移動して、【ENT】キーを押します。【F3】キーでグループ設定が可能です。

◀ ▶

○係数の設定テ ン キ ー を 用 い て 直接 入 力 し て く だ さい。符号の切替えには【F1】キー、指数入力への移動には【 】【 】キーを使用します。入力後、【ENT】キーを押します。

◀ ▶

注記

係数は表示値との関係を十分に把握したうえで設定してください。特に、指数の設定にはご注意ください。また、｢5-10 ページ 係数チェック｣にて、入力値と表示値の関係を確認することができます。

○表示桁の設定【▲】【▼】キ ー で 設 定 す る 表 示 桁 に カ ー
ソ ル （ 反 転 表 示 ） を 移 動 し て 、 【ENT】
キーを押します。

○単位の設定【▲】【▼】【 】【 】キーで設定する単位にカーソル（反転表示）を移動して、【ENT】
キーを押します。

◀ ▶

【F2 】【F3】キーで一覧表示のページを切
替えます。

5-8

５．３

係数・表示桁・単位❏ Cap/RO からの係数設定
変換器に添付されているデータシートには、定格容量(Cap)と定格出力(RO)を記載しています。この定格容量と定格出力、および単位を入力することで、係数の設定が可能です。
①係数・表示桁・単位の設定画面から、
Cap/RO を入力するチャンネルの係数にカーソルを移動し、 【F2 】キーを押します。

② 【▲】【▼】キ ー で 設 定 す る 項 目 に カ ー ソルを移動して、【ENT】キーを押します。
単位と表示桁の設定方法については、前述の通りです。

○定格容量の設定
テンキーで直接入力します。符号の切替
えは【F1】キー、小数点は【F2】キー、数値のクリアは 【F3】キーを使用します 。
入力後、【ENT】キーを押します。

○定格出力の設定
テ ン キ ー で 直 接 入 力し ま す 。 入 力 後 、【ENT】キーを押します。

○RO の単位設定【 】【 】キ ー で 設 定 す る 単 位 に カ ー ソルを移動して、【ENT】キーを押します。

◀ ▶

5-9

５．３ 係数・表示桁・単位③全ての項目を設定したら、【ESC】キーを押します。左の画面を表示します。【はい】にカーソルを移動して、【ENT】キーを押します。設定を適用し、係数・表示
桁・単位の設定画面に戻ります。

○Cap/RO からの係数設定例「定格容量(Cap)：5 kN、定格出力(RO)：2 mV/V のロードセルの場合」
左のように各パラメータを設定すると、係数が算出され、薄いグレーで表示します。
設定を適用すると、表示値は物理量直読になります。

❏ 係数チェック
入力値と表示値の関係から、設定した係数が適正であるかを確認します。
①係数・表示桁・単位の設定画面から、
係数チェックするチャンネルの係数にカーソルを移動し、 【F1 】キーを押します。

◀ ▶

② 【 】【 】キ ー で 入 力 値 に カ ー ソ ル を 移動して、【ENT】キーを押します。

③入力値に係数を乗じた値を表示します。

注記

5-10

係数を設定した際は、係数チェックを実行してください。入力ミスによる
測 定 データ関 連 のトラ ブ ルを未 然に防ぐことができます。

５．４

熱電対基準接点の設定

５．４ 熱電対基準接点の設定
熱電対を使用して温度を測定する場合、基準となるポイントを本器内部の温度センサとするか、外部に設けた基準接点（ゼロコンなど）とするかを選択します。
①プログラム設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【3】を押して、「熱電対基準接点の設定」に移行します。

② 【▲】【▼】キーで設 定 す る チ ャ ン ネ ル の設定値にカーソルを移動して、【ENT】キーを押します。【F3】キーでグループ設定が可能です。

◀ ▶

③【 】【 】キーで「内部接点」「外部接点」のいずれかにカーソルを移動して、【ENT】キーを押します。

✍ 本器は、内部に温度センサを内蔵しています。これは熱電対測定時に、内蔵の温度センサを基準接点として測定リニアライズするためです。精度は、JISZ8704 C 級測定方式に相当します。熱電対でより精度の高い測定をするときは、「外部接点」に設定し、外部に基準接点（ゼロコンなど）を設け測定してください。この場合の精度は、JIS Z8704 B 級に相当します。
基準零接点とした場合の接続例
通常の導線

熱電対

Ｈ

L

測温接点
H

Ｌ
0.0℃
外部基準接点（ゼロコンなど）

5-11

５．５

TEDS センサ

５．５ TEDS センサ
TEDS（Transducers Electronic Data Sheet）とは、荷重計や変位計などの変換器に容量や定格出力等の各パラメータを記録した IC チップを内蔵し、対応した測定器によりパラメータ
を読取ることで、設定ミスなく短時間に自動認識する機能です。

❏ センサ設定の読込み①TEDS 対応のセンサを NDIS コネクタに接続します。
②プログラム設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【4】を押して、「TEDS センサの読込み」に移行します。

③ 読 込 む チ ャ ン ネ ル に 【▲】【▼】キ ー で カー ソルを 移動 して、 【F1】キー を押し ます。

④読込んだ設定を一時的に表示します。

✍

正常に読込みが完了すると値を画面に表示します。容量、単位、定格出力以外を表示している場合、その表示と原因は次の通りです。
表示-------********
変化なし

注記

5-12

内容
タイムアウト(通信エラー)
サポートしていない TEDS センサ
TEDS に対応していないセンサ、または未接続○NDIS コネクタを装備した TEDS 対応センサのみ、読込みが可能です。
○設定の読込みが完了しても、設定内容は更新していません。設定を変更する場
合は次項の「読込んだ設定の反映」を行ってください。

５．６

測定モードの切替え❏ 読込んだ設定の反映①TEDS センサの読込みを実行します。

②読込んだ設定を反映する場合は、【F2】
キーを押します。

◀ ▶

③【 】【 】キーで｢はい｣にカーソルを移動して、【ENT】キーを押します。
読 込 ん だ 設 定 を 指 定の チ ャ ン ネ ル に 反映(確定)します。

５．６ 測定モードの切替え
測定モードを切替えます。 測定点数や外部に接続する専用の 5 点スイッチボックスなどに合わせて、設定してください。
測定モードの詳細については｢4-8 ページ 測定モードについて｣を参照してください。
①プログラム設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【5】を押して、「測定モードの切替え」に移行します。

②【▲】【▼】キーでカーソルを移動して、モードを選択します。
③【ENT】キーを押します。設定したモードのチャンネル設定画面を表示します。

5-13

５．７ メジャーとダイレクトの切替え

５．７ メジャーとダイレクトの切替え
測定値を表示する方法には初期値を差引いて表示するメジャー値（モニタ表示上「M」
または「m」が付きます）、初期値を差引かないダイレクト値（モニタ表示上「D」が付きます）があります。
①メニュー画面から、【▲】【▼】キーでカー
ソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【2】を押して、「メジャー/ダイレクトの切替え」に移行します。

② 【▲】【▼】キーで表 示 を 切 替 え る チ ャ ン
ネルの設定値にカーソルを移動して、【ENT】キーを押します。【F3】キーでグループ設定が可能です。

◀ ▶

③【 】【 】キーで「メジャー」「ダイレクト」のいずれかにカーソルを移動して、【ENT】キーを押します。

④切替えを確定し、メジャー/ダイレクトの切替え画面に戻ります。

5-14

５．８

自動測定

５．８ 自動測定
設定した時刻や時間間隔で測定データを記録します。

❏ インターバル測定とはインターバル測定とは、一定の時間間隔「インターバルタイム」で自動的に測定・記録する機能です。機能の概要を、3 ステップで 1 つのステップを 2 回実行する例で説明します。
1 つのステップは、時間設定と繰返し回数から構成され、ステップ 01 からステップ 02、ステップ 03 と実行します。最大 5 ステップまでのインターバル測定が可能です。
次の設定における測定タイミングは、下図のようになります。
ステップ
01
02
03

内容
00 時間 01 分 00 秒 間隔で測定
00 時間 04 分 00 秒 間隔で測定
00 時間 30 分 00 秒 間隔で測定

繰返し回数
2回
2回
2回

インターバル測定の概念図
ステップ 01ステップ 02ステップ 03

時間
0
スタート

1

2

6

10

40

70[分]

：インターバルタイマによる自動測定点

このほか、後述する機能として、ステップを実行開始時刻で制御する「リアルタイムスター
ト」、ステップのループ制御を行う「GOTO ステップ」があります。
注記○インターバル測定を開始後に、「日付・時刻の設定」は変更しないようご注意ください。
○インターバル測定中に、モニタ画面以外を表示しておくと、インターバル測定はできません。インターバル測定開始後は、モニタ画面へ戻してください。

❏ インターバルタイマの設定
一定の時間間隔「インターバルタイム」で自動的に測定・記録するための時間間隔、繰返し回数を設定します。
①メニュー画面から、【▲】【▼】キーでカー
ソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【3】を押して、「自動測定の設定」に移行します。

5-15

５．８ 自動測定②【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【1】を押して、「インターバル測定の設定」に移行します。

③ 【▲】【▼】キ ー で 設 定 す る ス テ ッ プ に カ
ー ソ ル （ 反 転 表 示 ） を 移 動 し て 【ENT】
キー、もしくは【F1】キーを押します。

④【▲】【▼】キーで｢タイマー｣にカーソルを移動して、【ENT】キーを押します。

⑤一旦、左のように内容を設定します。再度【ENT】キーを押します。
注記

設 定 内 容 の数 値 を変 更 する場 合 は【ENT】キー、設定内容そのものを変
更 す る 場 合 は 、 【 F1 】 キ ー を 押 し ます。

◀ ▶

⑥【 】【 】キーで入力桁を移動し、テンキー、もしくは【▲】【▼】キーで数値を入力します。【F3】キーで、回数に無限回を入力します。入力後、【ENT】キーを押します。
注記

1 秒から 99 時間 59 分 59 秒まで、1秒間隔で設定が可能です。

⑦ステップ 1 に動作を設定しました。左は｢10 分間隔で、10 回測定する｣という内
容です。

5-16

５．８

自動測定

⑧続いて、同様の手順でステップ 2 に｢インターバル終了｣を設定します。以上で、インターバル測定の設定は完了です。

✍

インターバル終了のステップを設定しなくても測定は可能です。

⑨インターバル測定中は、｢インターバル測定の設定｣で設定を確認できますが、設定の変更はできません。ステップ内容を変更、追加する場合は、インターバル測定を停止してください。

❏ リアルタイムスタートの設定測定するタイミングを、日付から秒単位まで詳細に設定できます。また、日、時、分の項に「＊＊」を選択することにより毎日、毎時、毎分の設定ができます。＊＊を入力した項より上位の項（分の項を＊＊にした場合の日、時の項）に数字を入力することはできません。
①「インターバル測定の設定 」に移行します。【▲】【▼】キーで設定するステップにカーソル（反転表示）を移動して【ENT】
キーを押します。
②【▲】【▼】キーで｢実時刻｣にカーソル（反転表示）を移動して、【ENT】キーを押します。
③一旦、左のように初期設定となります。
再度【ENT】キーを押します。

○毎時 00 分 00 秒に設定する例【 】【 】キーで入力桁を移動し、「時」に
カ ー ソ ル を 合 わ せ 【F3】キ ー を 押 し て 毎
時に設定します。
分、秒まで設定する場合はテンキー、もし く は 【▲】【▼】キ ー で 数 値 を 入 力 し ます。入力後、【ENT】キーを押します。

◀ ▶

注記

リアルタイムスタートは 1 度の測定で終了となります。回数の設定はありません。繰 返 し測 定 を行 う場 合 は次 頁を参照してください。

5-17

５．８ 自動測定○毎時 00 分 00 秒で繰返し測定を行う例
ステップ 02 に「ステップ番号 01 へ飛ぶ」を設定し、回数を無限回に設定します。この設定で繰返し毎時 00 分 00 秒に測定を行います。回数の部分に数字を設定することも
可能です。
注記

注記

GOTO ステップの設定をしていないと
繰返し測定を行 いません。GOTO ス
テップの設定 については「5-19 ペー
ジ GOTO ステップの設定 」を参照してください。

リアルタイムスタートで日に「＊＊」、かつ特定の時間を設定した場合には、測定開始のタイミングに注意してください。

例） 下記のように、0 時、6 時、12 時の測定を設定した場合
ステップ

内容

繰返し回数

01

＊＊日 00 時 00 分 00 秒 に測定--

02＊＊日 06 時 00 分 00 秒 に測定--

03＊＊日 12 時 00 分 00 秒 に測定--

04

ステップ 01 へ∞

この場合、0 時、6 時、12 時に測定をしますが、ステップ 01 0 時の測定を最初に行い、次いでステップ 02 6 時、ステップ 03 12 時の測定とステップ順に実行します。 つまり、リアルタ
イムスタートによる自動測定を開始した時刻が、6 時前、12 時前であってもステップ 01 を実行しないかぎり、6 時、12 時の測定はしません。

注記

リアルタイムスタートの日付として 29、30、31 日を設定した場合、該当する日付がな
い月はスキャニング測定をしません。

❏ Goto ステップとはGoto ステップは、指定したステップへ移動する機能です。インターバル測定を繰返し行う場合などに使用します。ステップ 01 からステップ 03 までのインターバル測定を実行した後、同様のステップを再度実行します。
例えば、下表のように設定した場合、
ステップ

内容

繰返し回数

01

01 時間 00 分 00 秒 間隔で測定

1回

0202 時間 00 分 00 秒 間隔で測定

3回

03ステップ 01 へ

2回

次のように動作します。
ステップ１
ステップ２
ステップ３

5-18

①
↓
②→③→④
↓
⑤

⑥
↓
⑦→⑧→⑨
↓
⑩

⑪
↓
⑫→⑬→⑭

５．８

自動測定

実際のインターバル測定のタイミングは、以下の通りです。
スタート

↓ ①
②
③
④ ⑥
⑦
⑧
⑨
ステップによる測定点┗━┻━━┻━━┻━━┻━┻━━┻━━┻━━┻-----1
2
2
2
1
2
2
2
インターバル時間❏ Goto ステップの設定①「インターバル測定の設定 」に移行します。【▲】【▼】キーで設定するステップにカーソル（反転表示）を移動して【ENT】
キーを押します。
②【▲】【▼】キーで｢Goto｣にカーソル（反転表示）を移動して、【ENT】キーを押します。
③一旦、左のように内容を表示します。再度【ENT】キーを押します。

◀ ▶

④【 】【 】キーでカーソルを移動し、テン
キー、もしくは【▲】【▼】キーでステップ
番号と回数を入力します。入力後、【ENT】キーを押します。

⑤ステップ 03 に動作を設定しました。この設定は前ページの設定例と同じ内容です。以上で Goto ステップの設定は完
了です。

5-19

５．８ 自動測定❏ インターバル測定の開始/停止
インターバル測定を開始する前に、データの記録先、または出力先（外部プリンタなど）を設定してください。
○開始① 自 動 測 定 の 設 定 画 面か ら 【▲】【▼】キ ー
でカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【2】を押して、「インターバル測定の開始/停止」に移行します。

◀ ▶

②【 】【 】キーで「Sleep On」「Sleep Off」のいずれかにカーソルを移動して、【ENT】キーを押します。
スリープ）とは、測 定 時 以 外 の✍ Sleep（
時間は電源を OFF にし、消費電力を抑える機能です。ただし、タイマーが 1
分 未 満 のステップでは、スリープは機能しません。また、設定した全ステップのタイマーが 1 分未満の場合、左の画
面となります（スリープ機能は使用できません）。詳細 は、｢5-21 ページ スリ
ープ機能｣を参照してください。

③インターバル測定を開始します。測定中
は、サブ LCD に測定中を表すインジケータが点灯します。
○停止「インターバル測定の開始/停止」に移行します。【 】【 】キーで｢はい｣にカーソルを移動して、【ENT】キーを押します。

◀ ▶

○停止せずに電源 OFF した場合
インターバル測定中にパワーキーを押した
場合この画面を表示します。
【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キーで選択します。
「インターバル停止」を選択した場合、インターバルを停止し電源 OFF 状態となります。
「インターバル続行」を選択した場合、次のインターバル時間までは、 スリープ状
態になりますが、次の測定 の 5 秒前に電源が立ち上がり、以降は Sleep OFF 状態でインターバルを続行し ます。
また、電源を OFF しない場合は 【ESC】キーを
押します。

5-20

５．８

自動測定

❏ スリープ機能
インターバル測定開始時に「Sleep On」を選択すると、スリープ機能が有効になります。
このスリープ機能により、測定時以外は主要な電源を遮断して、消費電力を抑えることができます。開始時の測定終了後、スリープ状態（サブ LCD のインジケータが点灯）となり、ディスプレイも消灯します。
インターバル測定時の、スリープによる動作タイミングを下図に示します。スリープは、
測定開始時刻までの時間間隔（下図Ⓑの期間）が 1 分以上の場合に機能します。この時間が1 分未満の場合、スリープは機能しません。
スリープ動作中は、測定開始時刻の直前に自動的に電源が入り、測定終了後再びスリープ（電源 OFF）となります。
Ⓑ
Ⓑ
Ⓐ

Ⓐ

電源ＯＮ
スリープ中

電源 OFF
禁止期間

スリープ中

電源 OFF
禁止期間

スリープ

測定終了
インターバル測定
スタート

測定開始時刻5 秒前

測定終了

インターバルによる
1 回目の自動測定開始

測定開始時刻5 秒前

測定終了

インターバルによる2 回目の自動測定開始○スリープ動作の解除
スリープ機能は、一時的に解除することができます。【POWER】キーで電源を ON にしてください。スリープ動作を一時的に解除し、データの確認、データの吸い上げ等の操作が可能になります。再び【POWER】キーで電源を OFF にすると、スリープ動作を再開します。スリープ動作を完全に解除するには、インターバル測定を停止してください。
○スリープ動作中の表示
スリープ動作から、自動で電源が ON した場合、左の画面を表示します。また、手動で電源を ON した場合は、モニタ画面を表示します。

注記○自動電源 ON から測定終了までの間（上図Ⓐの期間）は、 【POWER】 キーで電源を OFFにしないでください。それ以降自動で電源 ON になりません（手動で電源 ON にするまでイ
ンターバル動作が中断します）。
○データ確認等の操作は、次回測定時刻まで充分余裕のある場合に行ってください。
○スリープの一時的な解除中に次回測定時刻が近づいた場合、表示をモニタ画面に戻し、電源は ON のままにしてください。インターバルによる測定終了後、自動でスリープ状態になります。また 「Sleep On」の場合は、インターバル全ステップ終了後、電源 OFF 状態にな
ります。

5-21

５．９ 各種チェック

５．９ 各種チェック
センサに対するチェック機能について説明します。

❏ 絶縁チェック
入力端子に接続されたセンサと、供試体間の絶縁抵抗をチェックします。供試体と入力端子の E を、アース線などで導通してください。
①メニュー画面から、【▲】【▼】キーでカー
ソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【4】を押して、「各種チェック」に移行します。

②【▲】【▼】キーで ｢絶縁チェック｣にカーソル（反転表示）を移動して【ENT】キー、
もしくは、テンキーの【1】を押します。

③絶縁チェックを開始し、その結果を表示します。500MΩ以上の場合は、「＊」を表示します。

❏ 抵抗チェック
入力端子 A-B 間に接続したひずみゲージなどの抵抗をチェックします。
① 各 種 チ ェ ッ ク 画 面 から 【▲】【▼】キ ー で｢抵抗チェック｣にカーソル（反転表示）を移動して【ENT】キー、もしくは、テンキーの【2】を押します。
②抵抗チェックを開始し、その結果を表示
します。

5-22

５．９

各種チェック❏ ばらつきチェック
10 回測定を行い、その最大値と最小値の差を「ばらつき」として出力します。
① 各 種 チ ェ ッ ク 画 面 から 【▲】【▼】キ ー で｢ばらつきチェック｣にカーソル（反転表示）
を移動して【ENT】キー、もしくは、テン
キーの【3】を押します。
②【ENT】キーを押します。

③ばらつきチェックを開始し、その結果を表示します。

❏ リード線抵抗チェック
1 ゲージ法 3 線式のリード線抵抗分電圧の確認を行います。これは、1 ゲージ法 3 線式で測定したひずみを、完全なひずみ補正（Comet）にて表示する場合に必要なデータです。
① 各 種 チ ェ ッ ク 画 面 から 【▲】【▼】キ ー で｢リード線抵抗チェック｣にカーソル（反転表示）を移動して【ENT】キー、もしくは、テンキーの【4】を押します。
②リード線抵抗チェックを開始し、その結果を表示します。

❏ ブリッジ出力チェック
1 ゲージ法 3 線式および 2 ゲージコモンダミー法にて、現在のブリッジ初期不平衡値が分かるように、補正演算なしの場合のブリッジ出力電圧をチェックします。
① 各 種 チ ェ ッ ク 画 面 から 【▲】【▼】キ ー で｢ブリッジ出力チェック｣にカーソル（反転表示）を移動して【ENT】キー、もしくは、テンキーの【5】を押します。
②ブリッジ出力チェックを開始し、その結果を表示します。

5-23

５．１０ 測定補助設定❏ 係数設定チェック
モニタチャンネルに対して、入力値と表示値の関係から 、設定している係数が適正であるかを確認します。
① 各 種 チ ェ ッ ク 画 面 から 【▲】【▼】キ ー で｢係 数設 定チェック｣にカー ソル （反 転表示）を移動して【ENT】キー、もしくは、テンキーの【6】を押します。

◀ ▶

②【 】【 】キーで入力値にカーソルを移動して、【ENT】キーを押します。
③入力値に係数を乗じた値を表示します。
注記

係数を設定した場合は、係数チェックを実行してください。入力ミスによる測 定 データ関 連 のトラブルを未然に防ぐことができます。

５．１０ 測定補助設定
シンプルメジャーや Comet の設定、測定環境への対応など、測定の補助に関する設定について説明します。

❏ シンプルメジャーの設定①メニュー画面から、【▲】【▼】キーでカー
ソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【5】を押して、「測定に関する設定」に移行します。

② 【 ▲ 】 【 ▼ 】 キ ー で ｢ シ ン プ ル メジ ャ ー の 設定｣にカーソル（反転表示）を移動して【ENT】キー、もしくは、テンキーの【1】を押します。

5-24

５．１０

測定補助設定◀ ▶

③【 】【 】キーで「シンプルメジャー」にカーソルを移動して、【ENT】キーを押します。

④シンプルメジャーを示すマークを表示します。同時に、センサモードに対応した測定対象（左の場合ひずみ）を表示します。

○シンプルメジャーとは測定データに対するパラメータを測定チャンネル毎に設定する方法については、「 ５．３係数・表示桁・単位」で説明しましたが、ある固定のパラメータを一括して設定する機能があります。これは、設定したセンサモードに応じて、下表のパラメータを自動で設定する機能で、「シンプルメジャー」と呼びます。
シンプルメジャー設定の内容
センサモードひずみゲージおよび変換器
熱電対

＊1

係数

表示桁

単位 （表示例）+1.0000######

1με+1.0000#####．#

0.1℃＊1+1.0000#####．#

0.1℃

直流電圧（300mV）+1.0000###．###

0.001mV

直流電圧（30V）+1.0000##．####

0.0001V～300mV+1.0000###．###

0.001mV～30V+1.0000##．####

0.0001V

白金測温抵抗体

直流電圧（AUTO） ＊2＊1 センサモードに熱電対等の温度センサを選択すると、シンプルメジャーの設定に関係なく係数、表示桁、単位は自動で固定されます。
＊2 センサモードに直流電圧（AUTO）を選択すると、シンプルメジャーの設定に関係なく入力電圧によって、表示桁と単位を自動で切替えます。

5-25

５．１０ 測定補助設定❏ Comet の設定
1 ゲージ法 3 線式にてひずみを測定するときに、リード線による不要成分の誤差や、入力が大きい場合のひずみ値の誤差を補正して測定するための設定です。この機能を「 Comet」と称しています。Comet の詳細については、「１０．３ ひずみの完全な補正方法（Comet）」を参照してください。
① 測 定 に 関 す る 設 定 画面 か ら 、 【▲】【▼】
キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【2】を押して、「Comet の設定」に移行します。

◀ ▶

②【 】【 】キーで「使用しない」「Comet A」「Comet B」のいずれかにカーソルを移動して、【ENT】キーを押します。

注記

センサモードが 1 ゲージ法 3 線式の場合、「使用しない」を選択しても非直線補正をして表示します。 Comet を選択することで、より正確なひずみ値を得ることができます。

❏ 測定環境の電源周波数
バッテリで駆動する場合、使用する地域の商用電源周波数に合わせてください。測定値が不安定であったり、ばらつきやシフトが生じる場合、設定の変更により抑制することができます。
① 測 定 に 関 す る 設 定 画面 か ら 、 【▲】【▼】
キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【3】を押して、「測定環境の電源周波数」に移
行します。

◀ ▶

②【 】【 】キーで「50Hz」「60Hz」のいず
れかにカーソルを移動して、【ENT】キーを押します。

5-26

５．１０

測定補助設定❏ イニシャルインの許可
初期値の処理での、イニシャルインを禁止することができます。不用意なイニシャルインの実行を防止します。
① 測 定 に 関 す る 設 定 画面 か ら 、 【▲】【▼】
キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【4】を押して、「イニシャルインの許可」に移行
します。

◀ ▶

②【 】【 】キーで「禁止」にカーソルを移動して、【ENT】キーを押します。

❏ バーンアウトチェックの設定
熱電対の断線状況を常時チェックします。
① 測 定 に 関 す る 設 定 画面 か ら 、 【▲】【▼】
キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【5】を押して、「バーンアウトチェックの設定」に移行します。

◀ ▶

②【 】【 】キーで「する」にカーソルを移動して、【ENT】キーを押します。

5-27

５．１０ 測定補助設定

✎memo

5-28

第

６章

記録の設定
６.１ 記録の設定概要 ················································

6 - 2

６.２ データメモリ ·························································

6 - 3

６.３ CF カード ······························································

6 - 6

６.４ ファイルのコピー ·················································

6 -11

６.５ データメモリ・CF カードへの記録 ····················

6 -12

６．１ 記録の設定概要

６．１ 記録の設定概要
内蔵のデータメモリ、および CF カードからの読出しやコピーなど、データの管理に関す
る操作、設定について説明します。これらの操作・設定の構成を以下に示します。

【F3】キーを押してメニューへ移行します。【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【6】を押して、「記録」に移行します。さらに、カーソルの移動、もしくはテンキーにて設定項目を選択します。

6-2

６．２

データメモリ

６．２ データメモリ
データメモリの構造、内蔵のメモリに蓄えられたデータの読出しや消去、およびデータ
番号の設定などについて説明します。

❏ データメモリの構造
シングルチャンネルモードでは、最大 80,000 データを記録することが可能です。データ
メモリは、1 つのエリアから成り立っており、測定した順にデータを格納していきます。
1 つのデータは、チャンネル、測定時刻、測定データ、単位で構成しています。

記録

測定データ

80000

データメモリ○チャンネル○測定時刻○測定データ○単位

40000○データメモリ詳細記録可能なデータ数は最大80,000 データです。データ数が80,000 に達すると、サブ LCD に「M」を表示し、それ以上のデー
タは記録しません。チャンネルを切替えても、データの格納先は変わりません。パソコンに取込んだ後のデータ整理の際は、チャンネルでソートしてください。

0

リングバッファの設定を「On」にしている場合、データ数が 80000 に達すると、一番古いデータを捨てて、最新のデータを記録し続けます。

記録

測定データ

80000

40000

リングバッファの設定が「On」の場合、データ数が 80,000 に達しても、サブ LCD に「M」は点灯しません。

データメモリ0
一番古い測定データ

6-3

６．２ データメモリ❏ データの読出し①データメモリ画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【1】を押して、「デー
タの読出し」に移行します。

② カ ー ソ ル 位 置 の 番 号の デ ー タ を 表 示 します。データ番号の切替えは、【▲】【▼】
キーで増減、もしくはテンキーで直接入力してください。データ番号をクリアする場
合は、【F3】キーを押します。

❏ データメモリの設定
記録するデータのデータ番号、およびリングバッファ機能の設定を行います。
①データメモリ画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【2】を押して、「デー
タメモリの設定」に移行します。
②【▲】【▼】キー、もしくはテンキーを押して、設定項目を選択します。

○データ番号の設定
カ ー ソ ル 位 置 の デ ー タ 番 号 を 【▲】【▼】
キーで増 減 、もしくはテンキーで直 接 入 力
し、【ENT】キーを押します。

注記

6-4

記録データ数以上の番号は設定できません。直近のデータを消去して、データを取直したい場合は、現在値より小さい数字（取直したいデータ数分）を設定してください。また、
データ番号を 00000 に設定することは、データメモリのデータを消去することと同じであり、一旦、00000 に設定すると、元に戻すことはできません。

６．２

データメモリ○リングバッファの設定【 】【 】キー で カー ソル を移動 して 、「Off」「On」を選択します。

◀ ▶

○リングバッファ ON での繰返し測定
データメモリの残量がなくなり、2 周目
以降の記録になると、表示が左のように変わります。

❏ データメモリのデータ消去
データメモリに記録している測定データを消去します。
①データメモリ画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【3】を押して、「デー
タメモリのデータ消去」に移行します。

◀ ▶

②【 】【 】キーで「はい」にカーソルを移動して、【ENT】キーを押します。

❏ データメモリのプリンタ出力
データメモリに記録している測定データをプリンタに出力します。
①データメモリ画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【4】を押して、「デー
タメモリのプリンタ出力」に移行します。
②コピーするデータの範囲を指定します。
【 】【 】キー で カー ソル を移 動 して 、「From」「To」の数値を【▲】【▼】キーで増減、もしくは テンキーで直接入力 します。

◀ ▶

③設定が完了したら、【ENT】キーを押します。プリントを開始します。
④プリントが完了すると、データメモリ画
面に戻ります。

6-5

６．３

CF カード○プリンタフォーマット
シングルチャンネルモード
2012/10/21 16:51:51[D]00 +
107 ue2012/10/21 16:51:58[I]00 +
107 ue2012/10/21 16:52:01[M]00 +
0 ue

マルチチャンネルモード
2012/10/21 17:23:29[M]00 2 ue
[M]01 +
0 ue
[M]02 1 ue
[M]03 2 ue
[M]04 1 ue
2012/10/21 17:23:36[M]00 2 ue

６．３ CF カード
CF カードに蓄えたデータの読出しや消去、およびファイルの設定などについて説明します。

❏ データの読出し①CF カード画面から、【▲】【▼】キーでカー ソ ル （ 反 転 表 示 ） を 移 動 し て 【ENT】
キー、もしくはテンキーの【1】を押して、「データの読出し」に移行します。

② 【▲】【▼】キ ー で読 出 す デ ー タ の フ ァ イ
ルにカーソルを移動して、【ENT】キーを
押します。

6-6

６．３

CF カード③ 選 択 し た フ ァ イ ル のデ ー タ を 表 示 し ます。【▲】【▼】【 】【 】キーで、表示をスク
ロールします。【F2】キーでデータの最初、【F3】キーでデータの最後を表示します。

◀ ▶

④【ESC】キーを押すと、データの読出し
画面に戻ります。

❏ ファイル名・ファイル形式の設定
CF カードに記録するデータのファイル名、およびファイル形式を設定します。
①データメモリ画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【2】を押して、「ファイル名・ファイル形式の設定」に移行します。

②ファイル名の設定
dat 以降の 3 桁の数字で、ファイル名を付けてください。カーソル位置の数字を【▲】【▼】キーで増 減 、もしくはテンキーで直接入力します。
③ファイル形式の設定【F3】キーで「TDS」、「CSV1」、「CSV2」を切替えます。
④ フ ァ イ ル 名 と フ ァ イル 形 式 を 設 定 し た
後、【ENT】キーを押します。
○書込みフォーマット
シングルチャンネルモード・TDS フォーマット
2011/02/11 05:13:21
M00 +0000100 ue
END

シングルチャンネルモード・CSV1 フォーマット（表計算ソフトイメージ）
Date Time,CH,DATA
2011/02/11 05:15:15,00,+100

シングルチャンネルモード・CSV2 フォーマット（表計算ソフトイメージ）
Date Time,CH00
2011/02/11 05:15:25,

+100

6-7

６．３

CF カードマルチチャンネルモード・TDS フォーマット
2011/02/11 05:14:20
M00 +0000100 ue
M01 +0000123 ue
M02 +0000234 ue
M03 +0000345 ue
M04 +0000456 ue
END

マルチチャンネルモード・CSV1 フォーマット（表計算ソフトイメージ）
Date Time,CH,DATA,CH,DATA,CH,DATA,CH,DATA,CH,DATA
2011/02/11 05:15:45,00,+100,01,+123,02,+234,03,+345,04,+456

マルチチャンネルモード・CSV2 フォーマット（表計算ソフトイメージ）
Date Time,CH00,CH01,CH02,CH03,CH04
2011/02/11 05:15:55,
+100,
+123,

+234,

+345,

+456

○同名ファイル設定時設定したファイル名のファイルがすでに CF カード内に存在している場合、追記を確認するメッセージを表示します。
追記する場合は「はい」を選択します。「いいえ」を選択すると「ファイル名・ファイル
形式の設定」に戻ります。
「はい」を選択した場合、追記先のファ
イル内容をチェックします。65,000 行に達しているファイルであれば、追記ができないため、「ファイル名・ファイル形式の設定」に戻ります。

注記

6-8

CSV 形式のファイル操作に関する注意○1 つのファイルの最大行数は 65,000 行です。
○65,000 行に達しているファイルへの追記はできません。
○インターバル測定などによってデータファイル（datXXX）が 65,000 行を超えた場合、自動でファイルを作成します。ファイル名は XXX＋1、もしくは任意（XXX＋1 が存在している場合）となります。

６．３

CF カード❏ ファイル削除
CF カードに記録している測定データファイルを消去します。
①CF カード画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【3】を押して、「ファイル削除」に移行します。
②【▲】【▼】キーでカーソルを移動して、削除するファイルを選択し、【ENT】キーを
押します。

◀ ▶

③【 】【 】キーでカーソルを「はい」に移動して、【ENT】キーを押します。

❏ 設定ファイルの保存
現在の設定を CF カードに保存します。各チャンネルのセンサモードを始め、プログラム
設定、自動測定の設定、測定に関する設定、インターフェースの設定など各種の設定値を保存します。測定対象や状況に応じた設定を保存し、それを読込むことで、各種の設定を簡単に再現することができます。
①CF カード画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【4】を押して、「設定
ファイルの保存」に移行します。
②set 以降の 3 桁の数字で設定ファイル名を付けます。カーソル位置の数字を【▲】【▼】キーで増 減 、もしくはテンキーで直 接入力し、【ENT】キーを押します。

◀ ▶

③【 】【 】キーでカーソルを「はい」に移動して、【ENT】キーを押します。
④進行状況の表示とともに、保存を開始します。
⑤保存が完了すると、CF カード画面に戻
ります。

6-9

６．３

CF カード○同名ファイル設定時設定したファイル名のファイルがすでに CF カード内に存在している場合、上書きを確認するメッセージを表示します。上書きする場合は「はい」を選択します。

❏ 設定ファイルの読込み設定条件を保存したファイルを CF カードから読込み、その設定を反映します。
①CF カード画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの【5】を押して、「設定
ファイルの読込み」に移行します。
②【▲】【▼】キーでカーソルを移動して、読込むファイルを選択し、【ENT】キーを押
します。

◀ ▶

③【 】【 】キーでカーソルを「はい」に移動して、【ENT】キーを押します。
④進行状況の表示とともに、読込みを開始します。
⑤読込みが完了すると「設定ファイルの読込み」に戻ります。

❏ CF カードのフォーマット
CF カードを初期化します。
①CF カード画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、
もしくはテンキーの 【6】を押して、「CFカードのフォーマット」に移行します。

◀ ▶

②【 】【 】キーでカーソルを「はい」に移動して、【ENT】キーを押します。
③フォーマットが完了すると、「 ピポ」とブザー音が鳴り、CF カード画面に戻り
ます。

6-10

６．４

ファイルのコピー

６．４ ファイルのコピー
データメモリの測定データを CF カードにコピーします。
①記録画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、もしくはテンキーの【3】を押して、「ファイルのコピー」に移行します。

②コピーするデータの範囲を指定します。
【 】【 】キー で カー ソル を移動 して 、「 From」「To」、さ らにファ イル名の 数値を【▲】【▼】キーで増減、もしくはテンキーで直接入力します。

◀ ▶

③ファイル形式の設定【F3】キーで「TDS」、「CSV1」、「CSV2」を切替えます。
④設定が完了したら、【ENT】キーを押します。進行状況の表示とともに、コピーを開始します。
⑤コピーが完了すると、記録画面に戻ります。
注記

dtm のファイル名 XXX は、自動でインクリメントします。

○同名ファイル設定時設定したファイル名のファイルがすでに CF カード内に存在している場合、追記を確認するメッセージを表示します。
追記する場合は「はい」を選択します。「いいえ」を選択すると「ファイルのコピー」に
戻ります。

6-11

６．５ データメモリ・CF カードへの記録

６．５ データメモリ・CF カードへの記録
測定データの記録先を設定します。データメモリのみ、CF カードのみ、もしくは両方に設定が可能です。
①記録画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【4】を押して、「データメモリ・CF カードへの記録」に移行します。

②【▲】【▼】キー、もしくはテンキーにて、
詳細設定項目に移行します。

○データメモリへの記録
データメモリへ記 録 する場 合 は、【 】【 】
キ ーで カーソ ルを「 On」 に移動 して 、【ENT】キーを押します。

◀ ▶

○CF カードへの記録
CF カードへ記録する場合は、【 】【 】キーでカーソルを「On」に移動して、【ENT】
キーを押します。

◀ ▶

③選択した記録先を、画面左上に表示します。左はデータメモリと CF カード両方を選択している状態です。また、「DM」、「CF」のとなりの四角は各メモリに対する、空き容量を簡易的に示して
います。

6-12

第

７章

インターフェースの設定
７.１ インターフェースの設定概要 ···························

7 - 2

７.２ RS-232C の設定 ················································

7 - 3

７.３ データの出力 ······················································

7 - 4

７.４ データ出力形式の設定 ····································

7 - 5

７.５ 外部表示器の設定 ············································

7 - 6

７.６ プリンタの設定の注意 ···································

7 - 6

７.７ リモート測定について ········································

7 - 7

７．１ インターフェースの設定概要

７．１ インターフェースの設定概要
RS-232C の通信条件を始め、データの出力先・出力形式など、外部機器とのインターフェースに関する設定、遠隔地からのリモート測定等について説明します。これら各種設定の構成を以下に示します。

【F3】キーを押してメニューへ移行します。【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【7】を押して、「インターフェースの設定」に移行します。
さらに、カーソルの移動、もしくはテンキーにて詳細設定項目を選択します。

7-2

７．２

RS-232C の設定

７．２ RS-232C の設定
各種の専用ケーブル（別売）を用いてパソコンや外部プリンタと接続すれば、「TC-32K」のコントロールやデータの取込み、および測定データの印字等が可能となります。RS-232C
インターフェースの通信条件の設定について説明します。

❏ 通信条件の設定① イ ン タ ー フ ェ ー ス 設定 画 面 か ら 、 【▲】【▼】キーでカーソル（反転表示）を移動
し て 【ENT】キー、 も し く は テ ン キ ー の【1】を押して、「RS-232C の設定」に移行します。

②通信条件を表示します。
③【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーを押して、変更する設定項目を選択します。

④各項目の設定値は左の通りです。
【 ◀ 】【 ▶ 】キーで設定値を選択し、【ENT】
キーを押します。

注記

タイムアウトは 1～99 秒の範囲で設定できます。

7-3

７．３ データの出力

７．３ データの出力測定データをプリンタに印字したり、外部に接続した表示器などに出力します。

❏ 機器との接続
接続する機器によって、ケーブルが異なります。ケーブルの一方を RS-232C コネクタへ、
もう一方を各機器の該当コネクタに接続してください。詳細については、各機器の取扱説
明書を参照してください。

プリンタケーブル
CR-4511
外部プリンタDPU-S245
データケーブル
CR-4521
外部表示器
EDU-11

RS-232C ケーブル
CR-5532

パソコン❏ データ出力先の設定と出力方法
接続した機器に応じて、データの出力先を設定します。
①各種設定画面から、【▲】【▼】キーでカー
ソル（反転表示）を移動して【ENT】、もしくはテンキーの【2】を押して、「データ出力先の設定」に移行します。

②【 ◀】【 ▶】キーで出力先を選択し、【ENT】
キーを押します。

7-4

７．４

データ出力形式の設定○Off
外部に何も出力しない（接続しない）場合に選択します。
○プリンタ
プリンタ｢DPU-S245｣へデータを出力および印字する場合に選択します。自動測定中は、測定と同時にデータを印字します。また、任意のタイミングにて、【ENT】キーで印字することができます。ご使用の際は、｢DPU-S245｣の取扱説明書をよくご確認のうえ、本器との通信条件を設定してください。
○表示器
外部表示器｢EDU-11｣を使用する場合に選択します。
○手動パソコンへのデータ出力をキー操作で行う場合に選択します。シングルチャンネルモー
ドでは、【ENT】キーを押すごとに測定データを出力します。マルチチャンネルモード（傾斜計モード）では、【ENT】キーでのスキャニング測定を実行後、全チャンネルの測定デー
タを出力します。また、自動測定中は測定ごとにデータを出力します。

７．４ データ出力形式の設定
データの出力形式を設定します。
①インターフェースの設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、もしくはテンキーの【3】を押して、「データ出力形式の設定」に移行します。
②変更する設定項目を選択します。

③各項目の設定値は左の通りです。
【 ◀ 】【 ▶ 】キーで設定値を選択し、【ENT】
キーを押します。
各項目の詳細については、次項を参照してください。

○データフォーマット
CSV
データをカンマ(",")で区切って並べたファイル形式です。主に表計算ソフ
トやデータベースソフトがデータを保存するときに使う形式です。
TDS
弊社独自の ASCII 形式のファイルです（プリンタ印字とほぼ同様のイメージで記録します）。

7-5

７．５ 外部表示器の設定

○ヘッダ
有り

各測定データにヘッダ（D、M、m）の情報を付与して出力します。

○タイムデータ
有り
各測定データに日付と時刻の情報を付与して出力します。

７．５ 外部表示器の設定
弊社製表示器｢EDU-11｣を接続する場合の通信条件を設定します。
①インターフェースの設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】、もしくはテンキーの【4】を押して、「外部表示器の設定」に移行します。
②変更する設定項目を選択します。

③各項目の設定値は左の通りです。
【 ◀ 】【 ▶ 】キーで設定値を選択し、【ENT】
キーを押します。

７．６ プリンタの設定の注意
オプションのプリンタ｢DPU-S245｣を使用する場合、下記の様に設定します。
①「TC-32K」とプリンタの「ボーレート設定」「データビット設定」「パリティ設定」「ストップビット設定」を同じに設定にします。
② プリンタ｢DPU-S245｣側 FLOW CONTROL を RTS/CTS に設定③「TC-32K」側 「RS-232C の設定」→「フロー制御設定」を「ハードウェア」に設定
この設定を合わせていない場合は、プリンタへの印字が正しく行われません。

7-6

７．７

リモート測定について

７．７ リモート測定についてバッテリを電源として自動測定する場合は、通常はスリープ機能を使用します。これにパソコンを組合わせたり、モデム等の通信を介することにより、遠隔地から 「TC-32K」の電源を ON/OFF する、データを取込むなどの制御が可能となります。

❏ 構成○リモートパワーコントローラ ｢RPC-05A｣「TC-32K」とパソコン、あるいはモデムの間に組込み、コマンド送信による電源のON/OFF や長期測定時のソーラー発電などの制御を行うコントローラです。詳細については、リモートパワーコントローラ｢RPC-05A｣の取扱説明書を参照してください。

専用ケーブル
CR-5532

RPC-05A

RS-232C ケーブル｢CR-5321｣
またはモデムパソコン

外部バッテリ

ソーラーバッテリ❏ 機能
パソコンからのコマンド送信により、「TC-32K」を制御します。コマンドの詳細について
は、別冊の｢TC-32K インターフェース編｣取扱説明書を参照してください。
○電源 OFF
RS-232C より「ESC＋O＋F」コマンドを送信すると、電源が OFF します。
○電源 ON
RS-232C より「ESC＋O＋N」コマンドを送信すると、電源が ON します。

7-7

７．７ リモート測定について

✎memo

7-8

第

８章

その他の設定８.１ その他の設定概要 ············································

8 - 2

８.２ オートパワーオフの設定 ···································

8 - 3

８.３ バージョン情報 ···················································

8 - 4

８.４ 日付・時刻の設定 ··············································

8 - 4

８.５ 日本語/英語の切替え ······································

8 - 5

８.６ メンテナンス ·························································

8 - 6

８.７ 工場出荷時設定 ················································

8 -10

８．１ その他の設定概要

８．１ その他の設定概要
日付や時刻の設定、表示言語の切替えなどについて説明します。メンテナンスにおいては、ソフトウェアのバージョンアップが可能です。その他の設定の構成を以下に示します。

【F3】キーを押してメニューへ移行します。【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【8】キーを押して、「その他の設定」に移行します。さらに、カーソルの移動、もしくはテンキーにて詳細設定項目を選択します。

8-2

８．２

オートパワーオフの設定

８．２ オートパワーオフの設定
操作しない状態が続くと自動的に電源を OFF にする機能です。電池駆動の場合は、電源切忘れ時の電池消費を防止します。ただし、AC アダプタから電源を供給している場合は、
オートパワーオフは機能しません。
①その他の設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【1】を押して、「 オートパワーオフの設 定 」 に 移 行 し ます。

◀ ▶

②【 】【 】キーでカーソルを移動して、オートパワーオフ機能の有効「On」、無効「Off」を選択します。
③オートパワーオフ機能を有効（On）にした場合は、無操作状態から電源が OFFになるまでの時間を 1 分単位で設定できます。【▲】【▼】キーで増減、もしくはテンキーで直接入力します。設定可能範囲は 1～99 分です。
④設定が完了したら、【ENT】キーを押します。設定条件を確定し、その他の設定画
面に戻ります。
注記○インターバル測定中は、電源の供給形態に関わらず、スリープ機能の状態を優先します。よって、スリープ機能が OFF の場合は、オートパワーオフは機能しません。
○パワーオフ時間に 00 分を設定することはできません。入力値エラーとなります。

8-3

８．３ バージョン情報

８．３ バージョン情報
ソフトウェアのバージョンを表示します。
①その他の設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【2】を押して、「バージョン情報」に移行します。

②バージョン情報、およびシリアルナンバを表示します。【ESC】キーを押すと、その他の設定画面に戻ります。
※左の画面は表示例です。実際のバージョンとは異なる場合があります。

８．４ 日付・時刻の設定
日付と時刻を設定します。
①その他の設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【3】を押して、「日付・時刻の設定」に移行します。

◀ ▶

② 【 】【 】キ ー で カ ー ソ ル を 移 動 し な がら、年・月・日・時・分・秒それぞれを【▲】【▼】キーでの数値増減、もしくはテンキーで直接入力します。

③設定が完了したら、【ENT】キーを押します。日付と時刻を確定し、その他の設定画面に戻ります。

8-4

８．５

日本語/英語の切替え

８．５ 日本語/英語の切替え
表示言語を日本語もしくは英語に切替えます。
①その他の設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【4】を押して、「日本語/英語の切替え」に移行します。

◀ ▶

② 【 】【 】キーでカーソルを移動して、「日本語」「English」を選択します。
③【ENT】キーを押します。選択した言語にて、その他の設定画面に戻ります。

左の画面は「English」を選択した状態です。

8-5

８．６ メンテナンス

８．６ メンテナンス
LCD バックライトの点灯条件や明るさの設定、およびキー操作時のブザー音量を調整します。また、ソフトウェアのバージョンアップを行います。

❏ LCD バックライトの設定①その他の設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【5】を押して、「メンテナンス」に移行します。

②さらに、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】キー、もしくはテンキーの【1】を押して、「LCD バックラ
イトの設定」に移行します。

◀ ▶

③ 【 】【 】キーでカーソルを移動して、「常時 ON」、「常時 OFF」、「操作後のみ」のいずれかを選択します。
「操作後のみ」はキー操作後、約 10 秒
間点灯します。
④【ENT】キーを押します。LCD バックライ
トの点灯条件を確定し、メンテナンス画
面に戻ります。

8-6

８．６

メンテナンス❏ LCD バックライトの明るさ①メンテナンス画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【2】を押して、「LCD バックライトの明るさ」に移行します。

◀ ▶

② 【 】【 】キーでカーソルを移動して、
「暗」、
「中」、
「明」のいずれかの明るさを選択します。
③【ENT】キーを押します。LCD バックライ
トの明るさを確定し、メンテナンス画面に戻ります。

❏ コントラストの設定①メンテナンス画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【3】を押して、「コントラストの設定」に移行します。

◀ ▶

②【 】【 】キーでカーソルを移動して、コントラストを調整します。
③【ENT】キーを押します。LCD バックライトのコントラストを確定し、メンテナン
ス画面に戻ります。

8-7

８．６ メンテナンス❏ ブザー音量の設定①メンテナンス画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【4】を押して、「音量の設定」に移行します。

◀ ▶

②【 】【 】キーでカーソルを移動して、音量を選択します。
③【ENT】キーを押します。ブザー音量を確定し、メンテナンス画面に戻ります。

❏ バージョンアップ
本器のソフトウェアをバージョンアップします。予め、バージョンアップファイルが入っている CF カードをご用意ください。
① バ ー ジ ョ ン ア ッ プ ファ イ ル が 入 っ て い
る CF カードをスロットに挿入します。

②メンテナンス画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【5】を押して、「バージョンアップ」に移行します。

◀ ▶

③【 】【 】キーで、「はい」にカーソルを移動して、【ENT】キーを押します。左の画
面は表示例です。実際のバージョンとは異なる場合があります。

④バージョンアップを開始し、進捗状況を表示します。

8-8

８．６

メンテナンス⑤ 左 は バ ー ジ ョ ン ア ップ が 正 常 に 終 了 した場合の画面です。一旦、電源を OFFにしてから、再度 ON にします。

○バージョンアップの失敗
左 は バ ー ジ ョ ン ア ッ プに 失 敗 し た 場 合 の画面です。画面の指示に従って、操作して
ください。
注記

失敗した状況によっては、画面に何も表示しないことがあります。このような場合は、電源を ON にしてからしばらく
お待ちください。それでも何も表示 しない場合は、弊社までご連絡ください。

8-9

８．７ 工場出荷時設定

８．７ 工場出荷時設定
設定内容を出荷時の状態に戻します。
注記

RS-232C の設定、表示言語を変更している場合、｢工場出荷時設定｣を実行しても出荷時状態には戻りません。

❏ 出荷時設定の実行①その他の設定画面から、【▲】【▼】キーでカーソル（反転表示）を移動して【ENT】
キー、もしくはテンキーの【6】を押して、「工場出荷時設定」に移行します。

◀ ▶

②【 】【 】キーで、「はい」にカーソルを移動して、【ENT】キーを押します。

③ 同 時 に デ ー タ メ モ リの 内 容 を 消 去 す る
場合は「はい」に、消去しない場合は「いいえ」にカーソルを移動して、【ENT】キーを押します。
④設定内容を出荷時の設定に戻し、モニタ画面に戻ります。

8-10

８．７

工場出荷時設定❏ 出荷時設定の一覧
工場出荷時の設定内容一覧です。
モニタ
モニタチャンネル
測定値表示
測定モード
プログラム設定
センサモード係数
単位
表示桁
熱電対基準接点
自動測定の設定
インターバル測定の設定
測定に関する設定
シンプルメジャーの設定
Comet の設定
測定環境の電源周波数
イニシャル値測定の設定
バーンアウトチェックの設定
記録関連
データメモリ
CF カードインターフェース関連
RS-232C *初期化設定：以前値

データ出力先
データ出力形式

外部表示器
その他の設定オートパワーオフ設定
表示言語 *初期化設定：以前値
メンテナンス

0ダイレクト
シングルチャンネルモード
4GAGE
1.00000με######
内部接点
全ステップ設定なし
使用しない
Comet A
50Hz
動作
しない
データ番号
リングバッファの設定
ファイル名
ファイル形式

00000
On
dat000
CSV1

ボーレート
データビットストップビット
パリティ
フロー制御
タイムアウトOff
データフォーマット
ヘッダの有無タイムデータの有無
ボーレート
パリティ

9600
8bit
1
Non
X-ON/X-OFF
5秒

On、10 分日本語
LCD バックライト設定
LCD バックライト明るさ
コントラスト
音量

TDS
有り
有り
19200
Non

常時 OFF
中3
中

8-11

８．７ 工場出荷時設定

✎memo

8-12

第

９章

測定点数拡張９.１ 測定点数拡張の概要 ·······································

9 - 2

９.２ スイッチボックス ··················································

9 - 3

９.３ 傾斜計アダプタ ···················································

9 - 8

９．１ 測定点数拡張の概要

９．１ 測定点数拡張の概要
本器は、測定点数が 1 点の測定器ですが、オプションのスイッチボックス、および傾斜計
用アダプタ等を使用することにより、測定点数を拡張することができます。これにより、 5点あるいは 2 点の同時測定が可能となります。また、測定モードは前述のシングルチャン
ネルモードとは異なる、マルチチャンネルモード、傾斜計モードに切替えます。
各設定などの操作手順は、前述と重複しているものが多くあります。よって、詳細を省略している部分については、｢第４章 モニタ表示と測定｣、｢第５章 測定の設定｣を参照してください。

❏ 測定点数拡張機器
測定点数を拡張するためには、以下のオプションをご用意いただく必要があります。
測定点数拡張用オプション
品名
スイッチボックス
2 軸挿入型傾斜
計アダプタ

9-2

型名
CSW-5B
CSW-5B-05
IA-32
IA-33

測定点数
5点
2 点（1 点）

測定モード

備考

マルチチャンネルモード

端子台のみ（5 チャンネルモニタ可能）

NDIS コネクタ併用

傾斜計 2 軸（1 軸）モード

ケーブル接続NDIS 接続９．２

スイッチボックス

９．２ スイッチボックス｢CSW-5B｣は、専用の 5 点スイッチボックスです。測定モードをマルチチャンネルモードに設定することにより、5 点のスキャニング測定、および各測定点のモニタ表示が可能になります。最大で、スイッチボックス 10 台分（50 チャンネル）のパラメータと初期値を設定、記憶することができます。詳細については、｢CSW-5B｣の取扱説明書を参照してください。

❏ CSW-5B の接続｢CSW-5B｣に付属のケーブル｢CR-655｣を使用して、本器の NDIS コネクタに接続します。
CR-655

注記｢ CR-655 ｣以 外 の ケー ブ ル を使 用
しないでください。故 障 の原 因 となります。

CSW-5B❏ データメモリ
マルチチャンネルモードでは、約 29,400 回の測定が可能です。1 つのデータは、ボック
スナンバ、測定時刻、および 5 つのチャンネル、測定データ、単位で構成しています。

記録

測定データ

29400
データメモリ0○ボックスナンバ○測定時刻○チャンネル（0～4）○測定データ×5 点○単位×5 点

○データメモリの詳細
約 29,400 回の測定が可能です。ボックスを切替えても、データの格納先は変わりません。リングバッファの設定が｢On｣のとき、データメモリが容量上限に達すると、一番古いデー
タを捨てて、最新のデータを記録し続けます。パソコンに取込んだ後のデータ整理の際は、ボックスナンバ
およびチャンネルでソートしてく
ださい。

9-3

９．２ スイッチボックス❏ 測定モードの設定
測定モードをマルチチャンネルモードに設定します。設定手順の詳細は｢５.６ 測定モードの切替え｣を参照してください。
①モニタ画面から、「チャンネル設定」「測定モードの切替え」に移行します。
②｢マルチチャンネルモード｣に設定します。

❏ チャンネルの設定｢CSW-5B｣1 台につき 5 点（0～4 チャンネル）、最大 10 台分（50 点）までのチャンネルを設定することが可能です。
○チャンネルの表記
チャンネルは 2 桁の数字で表現しています。上位がボックスナンバ、下位が｢CSW-5B｣の各チャンネルとなります。
左は、ボックスナンバが 5 の｢CSW-5B｣
を接続中のモニタ画面です。

注記

マルチチャンネルモード選 択 時 は、接続中 の｢CSW-5B｣を自動 で認識して、モニタ表示を行います。未接続の場合
は、左の画面となります。ただし、未接続 の状 態 でも、各 チャンネルへのプログラム設定（センサモード、係 数等）は可能です。

❏ ボックスナンバの設定「CSW-5B」は本器でボックスナンバの設定します。
①モニタ画面から、【F2】キーを押して、「チ
ャンネル設定」に移行します。

9-4

９．２

スイッチボックス② モ ニ タ 画 面 か ら 、 【F1】キ ー を 押 し て 、「CSW-5B 設定」に移行します。

③ カ ー ソ ル 位 置 の 数 字 を 【▲】【▼】キーで増 減 、もしくはテンキーで直 接 入力 し、【F1】キーを押して「書込み」をします。
また、【F2】キーを押して「読込み」をすることで、「CSW-5B」に書込まれている ボックスナン バを読込 む ことができます❏ モニタ表示リアルタイムでモニタ表示するチャンネルは 1 点で、｢＊｣が点滅しているチャンネルがモ
ニタチャンネルです。モニタ測定時は、ボックスナンバを認識し、そのボックスナンバに対応したチャンネルのセンサモード、係数、表示桁、単位、初期値によって測定データを表示します。｢CSW-5A｣ではモニタチャンネルの切替えは手動、｢CSW-5B｣では自動でモニタチャンネルを切替えます。
○モニタチャンネルの切替え【▲】【▼】キーで｢＊｣を移動して、モニタするチャンネルを選択してください。
【ENT】キーを押すと、一度に 5 チャンネ
ル分のモニタ表示を更新することができます。

❏ モニタの種類の選択
シングルチャンネルモードと同様に、数値モニタ、波形モニタを選択することができます。
○モニタの種類の選択【 】【 】キーで「モニタ OFF」「数値モニタ」「波形モニタ」のいずれかにカーソルを移動して、【ENT】キーを押します。

◀ ▶

9-5

９．２ スイッチボックス○波形モニタ選択時
波形表示できるチャンネルは 1 点です。
｢波形モニタの設定｣画面のモニタチャン
ネルを、【 】【 】キーで選択します（ボックスナンバは選択できません）。

◀ ▶

❏ 初期値の処理
シングルチャンネルモードと同様に、イニシャルイン、初期値の書換えをすることができます。メジャーとダイレクトの切替えについても同様です。
○イニシャルイン
5 点の初期値を順に取ります。モニタチ
ャンネル以外は、スキャニング測定（モニタ）するまで初期値を表示しています。

○初期値の書換え「CSW-5B」が未接続の状態でも、初期値を書換えることができます。

❏ プログラム設定
センサモードを始めとする各種の設定手順は、シングルチャンネルモードの操作と同様です。熱電対の基準接点に関しては、｢CSW-5B｣1 台のボックス単位で設定します。また、マルチチャンネルモードでは、TEDS センサの読込みはできません。

注記

9-6

熱電対基準接点は、ボックス単位で設定します。

９．２

注記

スイッチボックス｢CSW-5B｣は、TEDS センサに対応していません。

❏ 測定値の記録「CSW-5B」の 0 から 4 チャンネルまで、順に測定と記録を行う方法をスキャニング測定と言います。このスキャニング測定には、【ENT】キーを押して測定・記録する手動測定、自動で記録を行う自動測定（インターバル測定）があります。
スキャニング測定時は、｢CSW-5B｣のボックスナンバを認識し、そのボックスナンバに対応したチャンネルのセンサモード、係数、表示桁、単位、初期値によって測定データを表示・記録します。手動、自動いずれも、操作手順はシングルチャンネルモードと同様です。

注記

測定値を記録する場合は、データメ
モ リもしくは CF カードへの記 録 を「On」にしてください。

✍【ENT】キーを押すと、一度に 5 チャ
ンネル分のモニタ表示を更新することができます。

❏ 各種チェック
センサのチェック、および係数の設定チェックを 5 点同時に行います。
○絶縁チェック、抵抗チェック【▲】【▼】キーで｢＊｣を移動して、チェック（モニタ）するチャンネルを選択してください。【ENT】キーを押すと、各チャンネルを順次チェックします。
○そのほかのチェック
チェック開始（【ENT】キーを押す）で、各チャンネルを順次チェックします。

9-7

９．３ 傾斜計アダプタ

９．３ 傾斜計アダプタ
別売の 2 軸挿入型傾斜計アダプタ｢IA-32｣を接続することにより、2 チャンネル（X 軸測
定値、Y 軸測定値）モニタと同時測定が可能になります。詳細については、｢IA-32｣の取扱説明書を参照してください。

❏ IA-32 の接続｢IA-32｣に付属のケーブル｢CR-65｣を使用して、本器と接続します。
IA-32

CR-65

｢CR-65｣の一方を本器の NDIS コネク
タへ、もう一方を｢IA-32｣の｢TC-32K｣
コネクタへ接続します。

挿入型傾斜計❏ データメモリ
傾斜計モードでは、1 軸はシングルチャンネルモードと同様、最大 80,000 データを記録することができ、2 軸は約 56,000 回の測定が可能です。1 つのデータは、測定時刻、および2 つのチャンネル（1 軸は 1 チャンネル）、測定データ、単位で構成しています。

記録
測定データ
（2 軸モード）

56000
データメモリ○測定時刻○チャンネル（n、n+1）○測定データ×2 点○単位×2 点0

9-8

○データメモリの詳細
傾斜計 2 軸モードでは、約 56,000回の測定が可能です。チャンネルを変えても、データの格納先は変わりません。リングバッファの設
定が｢On｣のとき、データメモリが容量上限に達すると、一番古いデータを捨てて、最新のデータを記録し続けます。パソコンに取込んだ後のデータ整理の際は、チャンネルでソートしてください。

９．３

傾斜計アダプタ❏ 測定モードの設定
測定モードを傾斜計 1 軸モード、もしくは傾斜計 2 軸モードに設定します。設定手順の詳細は｢4-8 ページ 測定モードの切替え｣を参照してください。
①モニタ画面から、「チャンネル設定」「測定モード切替え」に移行します。
②｢傾斜計 1 軸モード｣、もしくは｢傾斜計 2軸モード｣に設定します。

❏ チャンネルの設定
2 軸モードの場合、設定可能なチャンネルは、｢00-01｣、｢02-03｣のように偶数が 1 軸目となり、それに連続した数字で 2 軸目が自動で設定されます。1 軸モードは、00～19 の間で任意に設定可能です。
①カーソル位置の 1 軸目を 00 から 18 までの 偶 数 で 入 力 し 、 【ENT】キ ー を 押 し ます。

❏ モニタ表示
2 軸モードでは、2 つのチャンネルを同時にモニタ表示します。

注記

傾斜計モードでは、波形表示することはできません。

9-9

９．３ 傾斜計アダプタ❏ プログラム設定
傾斜計という変換器を使用するため、センサモードは 4 ゲージ固定となります。また、各種の設定手順は、シングルチャンネルモードの操作と同様です。

注記

設定可能なセンサモードは、左の 3
種類です。

注記

設定ができない項 目は、左 のように薄く表示します。

❏ 測定値の記録
2 軸モードでは、2 軸分（2 チャンネル分）の測定と記録を行います。【ENT/START】キー
を押して測定・記録する手動測定、自動で記録を行う自動測定があります。手動、自動い
ずれも、操作手順はシングルチャンネルモードと同様です。

注記

9-10

測定値を記録する場合は、データメ
モ リもしくは CF カードへの記 録 を「On」にしてください。

９．３

傾斜計アダプタ❏ 傾斜計モードでできないこと
傾斜計モードでは、次の操作や設定を行うことはできません。
○初期値の処理
イニシャルインモニタチャンネルのイニシャルイン
初期値の書換え○メニュー
メジャー/ダイレクトの切替え○プログラム設定
熱電対基準接点の設定
TEDS センサの読込み

9-11

９．３ 傾斜計アダプタ

✎memo

9-12

第

１０章

ひずみの補正方法
１０.１ ひずみゲージの結線方法 ·····························

10 - 2

１０.２ センサケーブルの延長による感度低下 ·····

10 - 3

１０.３ ひずみの完全な補正方法(Comet) ·············

10 - 7

１０.４ リード線抵抗の求め方 ···································

10 - 9

１０.５ 1 ゲージ 4 線式での補正方法 ·····················

10 -10１０．１ ひずみゲージの結線方法

１０．１ ひずみゲージの結線方法
測定可能なひずみゲージの結線法について解説します。次節には、ひずみゲージの延長による感度低下率を示します。

❏ 1 ゲージ 4 線式
4 本のリード線を用い、リード線抵抗によるゲージ率の補正や接触抵抗による測定誤差が生じない新たなひずみ測定法です。リード線による初期不平衡を除くことができ 、リード線の温度補償ができます。
1 ゲージ法 3 線式では可能な限り太く短い電線を推奨していますが、1 ゲージ 4 線式では細いリード線やコネクタによる結線、および延長が可能です。そのため配線作業の効率化と配線ミスの防止、リード線の再利用などひずみ測定のコストダウンが実現できます。

❏ 1 ゲージ法 2 線式（1/4 ブリッジ）
アクティブゲージ（ひずみを検出するゲージ）以外のブリッジ辺には、内蔵の固定抵抗を使用してブリッジ回路を構成します。
ひずみゲージまでのリード線抵抗は、アクティブゲージにのみ直列に入ります。細い線や長いリード線を使用すると初期不平衡が除けなかったり、温度変化による抵抗変化分がそのまま見かけひずみとして測定値に混入してきます。また、ひずみゲージの温度補償もできません。より精度良く測定する場合は 1 ゲージ法 3 線式をお勧めします。
端子台の B と C をショートすることで 1G3W のセンサモードで測定できます。ただし、本器では正式にサポートしていません。
注記

Comet B で測定してもリード線抵抗の自動補正は行いません。Comet A と同様に測定値を補正する必要があります。詳しくは、「１０．３ ひずみの完全な補正方法(Comet)」を参照してください。

❏ 1 ゲージ法 3 線式（1/4 ブリッジ）
接続に 3 平行線を使用することで、1 ゲージ法 2 線式におけるリード線の影響を補償します。リード線の抵抗がブリッジ辺の固定抵抗側にも入るため、リード線による初期不平衡の影響がなくなり、リード線の温度補償も可能です。ただし、ひずみゲージの温度補償はできません。

❏ 2 ゲージ法（1/2 ブリッジ）
ブリッジ回路の 2 辺に、内蔵の固定抵抗を使用します。ひずみゲージは同一ロットのものを使用し、接続には 3 平行線を使用します。ひずみゲージとリード線の温度補償が同時にでき、リード線による初期不平衡の影響がなくなります。

❏ 4 ゲージ法（フルブリッジ）
内蔵の固定抵抗を使用せずに、ブリッジの 4 辺ともひずみゲージで構成する方法です。変換器やトルク測定、または曲げ測定で出力を大きくしたい場合などに使用しますが、一般の応力測定では限られた測定方式です。この場合、ひずみゲージを接着した場所でブリッジを構成すれば、リード線を延長してもリード線による初期不平衡と温度の影響がなくな
ります。

10-2

１０．２

センサケーブルの延長による感度低下

１０．２ センサケーブルの延長による感度低下❏ 定電圧方式による測定
ひずみ測定では、ブリッジ印加電圧を一定とする定電圧方式を用いるのが一般的です。この方式によると、ひずみゲージ式変換器までのセンサケーブルを延長した場合、ケーブル
抵抗分の電圧降下により変換器に印加する電圧が低くなるため、感度の低下が起こります。
4GAGE でセンサケーブルを延長した場合の等価回路を下図に示します。
ｒ

Ｉ

A

Rg

E

ｒ
B

ｅ

Rg
ｒ
Ｅ
Ｉ
ｅ

: ブリッジの入力抵抗: ケーブル抵抗: ブリッジ印加電圧: ブリッジ印加電流: ブリッジ出力電圧

C※A、B、C、D はスイッチボックスのターミナルD

1 ゲージ法 2 線式、2 ゲージコモンダミー法、4 ゲージ法の場合、リード線による感度低下を含まない真値は下記の式により求められます。

真値 ＝ 測定値×

Rg＋2ｒ
Rg

1 ゲージ法 3 線式、2 ゲージ法（3 線式）の場合、1 辺に入るリード線抵抗が片辺であるため下記の式により求められます。

真値 ＝ 測定値×

Rg＋ｒ
Rg

1 ゲージ法 3 線式では、補正方式として Comet B を設定すると、本器でリード線抵抗を測定し、補正した正確な測定ができます。
CometA、B を設定した場合、イニシャル値を含んだ補正を行っているため、上記の補正計算は適切ではありません。詳しくは、「１０．３ ひずみの完全な補正方法(Comet)」を参照してください。

10-3

１０．２ センサケーブルの延長による感度低下○各種ケーブル抵抗値と定電圧時の感度低下
下表に、通常使用するセンサ延長ケーブルの抵抗値を示します。
ケーブル直流抵抗値（25℃）
２

断 面 積 （ mm ） 素 線 直 径 （mm）×素 線 数0.08
0.12× 70.14
0.12×12
0.23
0.12×20
0.3
0.18×12
0.34
0.12×30
0.5
0.18×200.75
0.18×30

10m あたりの往 復 の抵 抗 値 （Ω）
4.4
2.46
1.6
1.19
1.06
0.71
0.48

○定電圧モード、350Ω変換器を 0.5mm ２ケーブルで 1km 延長した場合
Rg
Rg＋2r

＝

350
350＋71＝ 0.831353....

0.83となり、約 17％の感度低下を生じます。更に長い延長で温度変化の激しい場合には、銅の温度係数（＋0.004/℃）による抵抗値の変化にも注意が必要です。

10-4

１０．２

センサケーブルの延長による感度低下❏ 定電流方式による測定
定電流方式では、印加電流を一定に保つため、ケーブル抵抗が変化しても変換器に印加 す
る電圧は一定であり、感度低下を無視できます。
電圧を一定とした定電圧方式の場合のブリッジ出力電圧ｅは、
ｅ＝ERg・k・ε・()
4
Rg＋2rとなり、ケーブル延長がない場合 （r＝0） と比べて×（

Rg） の出力低下となります。
Rg＋2r

次に電流 I を一定にした定電流方式の場合には、
ｅ＝

I・Rg
4

・k・ε

となり、延長ケーブルの抵抗 r による影響は計算上現れません。しかし、実際のセンサ
ケーブル延長では、抵抗（R）成分にキャパシタンス（C）や、インダクタンス（L）等の成分が加わるため、定電流方式でも、本器の様にパルス印加型のデータロガーでは若干の感度低下が発生します。 またブリッジ入力抵抗 Rg に比例して入力端の電圧が変化するので、厳密には使用する変換器の入力抵抗値によって補正が必要です。

方式の違いによる特徴（C,L の影響を除く）
方式
定電圧
方式

ケーブル抵 抗 による感 度 低 下
0.5mm ２ケーブルで
約 -17％/km（入 力 抵 抗 350Ω）

変 換 器 入 力 抵 抗 の影 響
無 視 できる
必 要 に応 じて補 正 する

定電流方式

無 視 できる

真値＝測定値×

公称 入 力抵 抗（350Ω）
実 際 の入 力 抵 抗

本器の定電流方式のセンサモードは、「4GAGE C350Ω」です。その他のひずみ測定センサ
モードは全て定電圧方式です。定電流方式のセンサモードは、公称入力抵抗 350Ωの 4 ゲー
ジ式変換器を接続した場合に正規の感度となる様に校正しています。350Ω以下の変換器を接続すると測定値は低く表示し、350Ω以上の場合は高く表示します。
また、延長したケーブル分を含めた入力抵抗が 750Ω（標準分解能モード）以上になるとブリッジ電源が飽和し、測定不能となります（断面積 0.5mm２のケーブルで、約 5.6km）。
変換器の入力抵抗が 350Ωより大きくずれている場合、次式により補正します。
真 値 ＝測 定 値 ×

公称入力抵抗実際の入力抵抗○変換器入力抵抗 355Ωの場合
真 値 ＝測 定 値 ×

350
355

≒測 定 値 ×0.986となりますので、測定器側の係数に 0.986 を設定します。
当社のセンサは通常、入出力コード付であるため、試験成績書に記載の入力抵抗や感度等の諸性能は、その入出力コード端での値となっています。

10-5

１０．２ センサケーブルの延長による感度低下○実測例
使用センサケーブル：k-PNCT-SB(1)4×0.5SQ （0.5m ２ 4 心キャプタイヤ）
導体抵抗：70mΩ/m（往復）センサ：350Ωブリッジ
ケーブル延長距離[ｍ]
100

1000

2000

3000

4000

5000

定電流 350Ω

感

95

度
[％]
90

注記

10-6

定電圧 350Ω

上 記 の例 以 外 のケーブルを使 用 する場 合 、特 に浮 遊 容 量 の大 小 により、感 度 低
下 の度 合 は異 なります。

１０．３

ひずみの完全な補正方法（Comet）

１０．３ ひずみの完全な補正方法（Comet）
Comet：Complete Compensation Method of Strain の略称❏ 1 ゲージ法 3 線式での補正方法
ひずみゲージを用いてひずみ測定をする場合、通常ひずみゲージをブリッジ構成辺の一部とするホイートストンブリッジを組み、その抵抗変化を測ります。
ε
R＋ ΔR
eｒ

ｒ

ｒ

B
R2

ｒ

C
A
R4

R3
D

出力
ｅ
ｅ０

ｅ in
ｅ
ｅ０
K
ε
R
ΔR
ｒ
ｅｒ

：ブリッジ電源電圧：ひずみ発生時のブリッジ出力電圧：初期不平衡時のブリッジ出力電圧：ゲージ率：供試体のひずみ：ゲージ抵抗：ひずみにより変化した抵抗分：リード線抵抗：リード線抵抗の両端電圧

R2
R3
ダミー抵抗
R4
ただし、R3=R4

電源
ｅ in

上図のようにブリッジの AB 辺がアクティブゲージである場合、出力電圧ｅはひずみ εに完全に比例しません。本器は誤差を自動的に補正するために 3 種類の演算を用意しています（選択方法は、「５．１０ 測定補助設定」を参照してください）。センサモードの設定が 1
ゲージ法 3 線式の場合にのみ補正演算を実行します。
また、「TC-32K」はブリッジ電源電圧が 1V の仕様となっていますが、校正器や他の測定器との互換を取るため、ブリッジ出力電圧に対して補正をかけています。

注記

CometA 補正が可能なチャンネルのデータはヘッダが「ｍ」になります。

10-7

１０．３ ひずみの完全な補正方法（Comet）❏ Comet NON
従来の補正法で、以下の補正演算を行います。
2ｅε＝ ----- ・ ------------------ｅin
K―――― －ｅ2

ゲージ率 K は 2.00 として補正演算を行います。
ダイレクト測定時は初期不平衡値を差引かないひずみ値 εD を、ヘッダを「D」として演算を行い出力します。
メジャー測定時は、初期不平衡値を差引いたひずみ値 εM を、ヘッダを「M」として下記の演算を行い出力します。

❏ Comet A
ブリッジの初期不平衡による非直線性誤差を補正する補正方法で、初期不平衡値が大きい場合に有効です。初期不平衡によるブリッジ出力電圧ｅ ０ をイニシャルイン時に測定し内部に記憶しています。測定時にひずみ発生状態のブリッジ出力電圧ｅを 測定し以下の補正演
算を行います。
ｅ－ｅ ０ｅinε＝------- ・ -----------------------------------------------ｅinｅin －ｅ）・（ -------K（ ------＋ｅ ０）22

ゲージ率 K は 2.00 として補正演算を行います。
測定時は初期不平衡値を差引いた値 εm を、ヘッダを「m」として演算を行い出力します。
Comet A 演算を行うと、上記式からイニシャルイン時に記録した、初期不平衡値を含む補正演算を行うため、メジャー測定のみになります。測定を開始する初期不平衡状態で、イニシャルインを行ってから測定してください。

❏ Comet B （1 ゲージ法 3 線式のみ）
Comet A の補正方法と同時に、リード線による感度低下を補正する場合に使用します。初期不平衡時のブリッジ出力電圧ｅ ０ 、リード線抵抗の両端電圧ｅ ｒ をイニシャルイン時に測定し内部に記憶しています。測定時にひずみ発生状態のブリッジ出力電圧ｅを測定し以下の補正演算を行います。
ｅ－ｅ ０
ｅin
ε
＝ -------- ・ -----------------------------------------------Kｅinｅin（ ------ －ｅ ）・ （------- ＋ｅ ０ －ｅ ｒ ）22

ゲージ率 K は 2.00 とし補正演算を行います。
測定時は初期不平衡値を差引いた値 εm を、ヘッダを「m」として演算を行い出力します。
Comet B 演算を行うと、式からイニシャルイン時に記録した、初期不平衡値、リード線抵抗の両端電圧を含む補正演算を行うため、メジャー測定のみになります。測定を開始する
初期不平衡状態で、イニシャルインを行ってから測定してください。

10-8

１０．４

リード線抵抗の求め方

１０．４ リード線抵抗の求め方
1 ゲージ法 3 線式を使用している場合、リード線抵抗データリストと、リード線抵抗チェ
ックの値から、イニシャルイン時のリード線抵抗値、現在のリード線抵抗値を下記の式から求めることができます。
リード線抵抗データリストのデータｅ ｒ 0 と、イニシャルリストのデータｅ ０ からイニシャ
ルイン時のリード線抵抗値ｒ ０ を次の式で求めることができます。
ｒ 0 ：イニシャルイン時のリード線抵抗値（Ω）
ｅ ｒ0 ：イニシャルイン時のリード線抵抗電圧（μＶ）
ｅ 0 ：イニシャルイン時のブリッジ出力電圧（μＶ）
Rg ：ブリッジ抵抗｢ダミー抵抗の抵抗値 120Ω、240Ω、350Ω」
ｒ０ =

ｅ ｒ０
１＋ｅ ０ -ｅ ｒ０

×Rg

ダイレクトチェックのデータｅと、リード線抵抗チェックのデータｅ ｒ から現在のリード線抵抗値ｒを次の式で求めることができます。
r：リード線抵抗値（Ω）

e ｒ ：リード線抵抗チェック時のリード線抵抗の両端電圧（μＶ）
e：ダイレクトチェック時のブリッジ出力電圧（μＶ）
Rg ：ブリッジ抵抗「ダミー抵抗の抵抗値 120Ω、240Ω、350Ω」
ｒ =ｅｒ
１＋ｅ-ｅ ｒ×Rg

Comet B の補正計算は、イニシャルイン時のリード線抵抗を、記録し補正しています。現
在のリード線抵抗をもとにしたひずみεを求める場合は次の式で求めることができます。
ｅ ｒ０
ｅ０
ｅｒ
ｅ

：イニシャルイン時のリード線抵抗電圧（μＶ）：イニシャルイン時のブリッジ出力電圧（μＶ）：リード線抵抗チェック時のリード線抵抗の両端電圧（μＶ）：ダイレクトチェック時のブリッジ出力電圧（μＶ）
ｅ・（１－ｅ ｒ０ ）－ｅ ０ ・（１－ｅ ｒ）

ε＝

注記（１＋ｅ ０ －ｅ ｒ０ ）・（１－ｅ－ｅ ｒ ）○Comet A、B を使 用 して測 定 するには、補 正 式 のようにｅ ０ （初 期 不 平 衡 値 ）、ｅｒ（リード線抵 抗の両端 電圧）を測 定するためイニシャルインを行う必要 があります。イニシャルインにより初期値、リード線抵抗値（Comet B）を測定します（イニシャルインを実行しないとｅ ０ 、ｅ ｒ が測定しないため、正しく測定できません）。
○Come t A 、B 設 定 時 の 測 定 は、補 正 演 算 を行 っ ているた め、センサモ ードが1 ゲージ法 3 線式に設定しているチャンネルは、自動的にメジャーモードに設定し
ます。

10-9

１０．５ １ゲージ４線式での補正方法

１０．５ １ゲージ４線式での補正方法○測定原理
1 ゲージ 4 線式ひずみ測定方法は、ホイートストンブリッジ回路を使用せず、ゲージ抵
抗(R)と基準抵抗(Rs)による簡単な直列回路を構成しひずみを測定します。ゲージ抵抗で発生する電圧(V)と基準抵抗で発生する電圧(Vs)からひずみを求めます。
電流の流れる経路と電圧を測定する経路が別なので、リード線抵抗または、接触抵抗(r)の影響を受けずに測定することができます。

R：ゲージ抵抗ｒ1

Rs：基準抵抗R

r 1 ～r 4 ：リード線抵抗

ｒ2
ｒ3

V

ｉ：ゲージ抵抗と基準抵抗に流れる電流
E：電圧源
E

ｒ4

V：ゲージ抵抗に発生する電圧
V S ：基準抵抗に発生する電圧

ｉ
Rs

Vs

○補正演算
1 ゲージ 4 線式ひずみ測定方法で測定する場合、初期不平衡が大きくなると上記(V)に発生する出力電圧ｅはひずみ εに完全には比例しません。本器は誤差を自動的に補正するために演算を用意しています。
上記(V)に発生する初期不平衡時の出力電圧ｅ ０ を、イニシャルイン時に測定し内部に記憶します。測定時にひずみ発生状態の出力電圧ｅを測定し以下の補正演算を行います。
ｅ－ｅ ０εｍ ＝ -----------------１＋2・ｅ ０○補正演算の設定
補正演算は演算設定と連動して、補正演算する、しないを選択できます。
Comet 設定
Comet NON
Comet A
Comet B

補正演算
しない
する
する

「しない」を選択した場合は、出力電圧ｅがそのまま出力します。

センサモードは、ゲージ抵抗により「1G4W 120」「1G4W 240」「1G4W 350」に設定して
ください。

10-10

１１章

第

仕様
１１.１ 仕様 ··············································································

11 - 2

１１.２ 標準付属品 ·······························································

11 - 6

１１.３ オプション ···································································

11 - 7

１１.４ 外観図 ·········································································

11 - 8

１１．１ 仕様

１１．１ 仕様■ 測定点数
1点

単体

NDIS コネクタ、またはワンタッチ端子からのセンサ入力

5点

CSW-5A・B 組合せ時

CSW-5A、CSW-5A-05、CSW-5B、CSW-5B-05■ 適用センサ

ひずみ測定

熱電対温度
測定

電圧測定
白金測温抵抗
体温度測定

1G4W 120Ω

1 ゲージ 4 線式 120Ω

1G4W 240Ω

1 ゲージ 4 線式 240Ω

1G4W 350Ω

1 ゲージ 4 線式 350Ω

1G3W 120Ω

1 ゲージ法 3 線式 120Ω

1G3W 240Ω

1 ゲージ法 3 線式 240Ω

1G3W 350Ω

1 ゲージ法 3 線式 350Ω

2GAGE

2 ゲージ法120～1000Ω

4GAGE

4 ゲージ法120～1000Ω

4G C350

4 ゲージ定電流法 350Ω

4GAGE 0-2V

4 ゲージ法 0-2V
120～1000Ω

熱電対 T(CC)

熱電対 T

熱電対 K(CA)

熱電対 K

熱電対 J(IC)

熱電対 J

熱電対 B

熱電対 B

熱電対 S

熱電対 S

熱電対 R

熱電対 R

熱電対 E（CRC）

熱電対 E

熱電対 N

熱電対 N

DC 300mV

電圧測定 ±300mVDC 30V

電圧測定 ±30V

DC AUTO

電圧測定 ±30V ※1

Pt100 3W

白金温度測定 3 線式

※1 本体からの 1 チャンネル測定のみ

11-2※ 1 ゲージ 4 線式はオプショ
ンのアダプタを使用
ブリッジ電圧DC1V 44ms（50Hz）※ ブリッジ電圧 DC2V
24ms（50Hz)

リニアライズ
デジタル演算JIS C1602-1995

入力インピーダンス
V 1/1
500ＭΩ以上
V 1/100
1ＭΩ以上
リニアライズ デジタル演算
JIS C1604-1997 Pt100

１１．１

仕様

■ 測定範囲
測定項目

レンジ
×1

ひずみ測定

直流電圧
測定

×10

×1
×10
×1
×10

熱電対
温度測定

白金測温抵抗
体温度測定

---

---

測定モード

初期値記憶
範囲

イニシャル
ダイレクト
メジャー

±160000
×10－６ひずみ

測定範囲±30000×10 －６ひずみ±300000
×10 －６ひずみ
V1/1± 30.000mV± 300.000mV
V1/100± 3.0000 V± 30.0000 V
T：-250～＋ 400℃
K：-210～＋1370℃
J：-200～＋1200℃
B：＋200～＋1760℃
S：- 10～＋1760℃
R：- 10～＋1760℃
E：-210～＋1000℃
N：-200～＋1300℃-200～＋850℃

イニシャル
ダイレクト
メジャー

サンプリング
速度V1/1±160.000mV
V1/100±16.0000 V

ダイレクト-----------

ダイレクト-----------

80ms(50Hz 地域)
67ms(60Hz 地域)※ 4 ゲージ法 0-2V モード(差動トランス型変位計等、4GAGE のみ)の測定範囲は
×1:±15000×10-6 ひずみ、×10:±150000×10 -6 ひずみとなります。

■ 測定確度センサモード

レンジ

分解能

確 度
（23℃±5℃）

確度の温度係数(％rdg/℃)

確度の経年変化(％rdg/年)

ひずみ（1 ゲージ 4 線 式
を除 く）

×1
×10

1×10 －６ひずみ
10×10 －６ひずみ±(0.08％rdg＋1digit)±(0.08％rdg＋1digit)±0.002±0.002

±0.02
±0.02

ひずみ
（1 ゲージ 4 線 式 ）×1×10

1×10 －６ひずみ
10×10 －６ひずみ±(0.28％rdg＋1digit)±(0.28％rdg＋1digit)±0.002±0.002

±0.02
±0.02

電 圧 V1/1

×1×100.001mV
0.010mV±(0.08％rdg＋3digit)±(0.08％rdg＋3digit)±0.0024±0.0024

±0.02
±0.02

電 圧 V1/100

×1×10

0.0001V
0.0010V±(0.08％rdg＋2digit)±(0.08％rdg＋2digit)±0.002±0.002

±0.02
±0.02

白金測温抵抗体
Pt100 3W

---

0.1℃

±(0.08％rdg＋0.3℃)±0.002±0.05※ レンジは自動切換え

リード線抵抗補正
CometB(1 ゲージ法 3 線式)

ゲージ抵抗

リード線抵抗値補正範囲120Ω

約 100Ω以下240Ω

約 200Ω以下350Ω

約 300Ω以下11-3

１１．１ 仕様

熱電対温度測定
種 類

測定範囲

分解能

T

- 250～- 200℃
- 200～- 100℃
- 100～＋ 400℃

0.1℃
0.1℃
0.1℃

K

- 210～- 160℃
- 160～
0℃
0～＋ 960℃＋ 960～＋1370℃

0.1℃
0.1℃
0.1℃
0.1℃

J

- 200～- 160℃
- 160～
0℃
0～＋ 700℃＋ 700～＋1200℃

0.1℃
0.1℃
0.1℃
0.1℃

B

＋ 200～＋ 280℃
＋ 280～＋ 800℃
＋ 800～＋1760℃

S

確 度 (23℃±5℃)
外部基準接点

内部基準接点±( 0.38 ％rdg ＋0.6 ℃)±( 0.15 ％rdg ＋0.2 ℃)±( 0.10 ％rdg ＋0.2 ℃)±( 0.19 ％rdg ＋0.3 ℃)±( 0.12 ％rdg ＋0.2 ℃)±( 0.08 ％rdg ＋0.1 ℃)±( 0.10 ％rdg ＋0.9 ℃)±( 0.38 ％rdg ＋3.9 ℃)±( 0.15 ％rdg ＋1.4 ℃)±( 0.10 ％rdg ＋0.8 ℃)±( 0.19 ％rdg ＋1.6 ℃)±( 0.12 ％rdg ＋1.0 ℃)±( 0.08 ％rdg ＋0.5 ℃)±( 0.10 ％rdg ＋1.4 ℃)

0.5～0.4℃
0.3～0.1℃
0.1℃±( 0.16 ％rdg ＋0.2 ℃)±( 0.12 ％rdg ＋0.1 ℃)±( 0.08 ％rdg ＋0.1 ℃)±( 0.08 ％rdg ＋0.6 ℃)±( 0.04 ％rdg ＋4.0 ℃)±( 0.04 ％rdg ＋1.2 ℃)±( 0.05 ％rdg ＋0.4 ℃)±( 0.16 ％rdg ＋1.2 ℃)±( 0.12 ％rdg ＋0.8 ℃)±( 0.08 ％rdg ＋0.5 ℃)±( 0.08 ％rdg ＋0.9 ℃)±( 0.04 ％rdg ＋4.0 ℃)±( 0.04 ％rdg ＋1.2 ℃)±( 0.05 ％rdg ＋0.4 ℃)- 10～＋ 200℃
＋ 200～＋1760℃

0.1℃
0.1℃

±( 0.09 ％rdg ＋0.6 ℃)±( 0.07 ％rdg ＋0.4 ℃)±( 0.09 ％rdg ＋1.2 ℃)±( 0.07 ％rdg ＋0.7 ℃)

R- 10～＋ 150℃
＋ 150～＋1760℃

0.1℃
0.1℃

±( 0.09 ％rdg ＋0.7 ℃)±( 0.07 ％rdg ＋0.4 ℃)±( 0.09 ％rdg ＋1.2 ℃)±( 0.07 ％rdg ＋0.7 ℃)

E- 210～＋ 550℃
＋ 550～＋1000℃

0.1℃
0.1℃

±( 0.17 ％rdg ＋0.2 ℃)±( 0.09 ％rdg ＋0.4 ℃)±( 0.17 ％rdg ＋1.4 ℃)±( 0.09 ％rdg ＋0.8 ℃)

N- 200～
0℃
0～＋1090℃＋1090～＋1300℃

0.1℃
0.1℃
0.1℃

±( 0.18 ％rdg ＋0.4 ℃)±( 0.08 ％rdg ＋0.2 ℃)±( 0.08 ％rdg ＋0.9 ℃)±( 0.18 ％rdg ＋1.6 ℃)±( 0.08 ％rdg ＋0.6 ℃)±( 0.08 ％rdg ＋1.2 ℃)※ センサの確度は含まず、また、熱電対 B は基準接点を使用しないチェック機能
測定範囲

絶縁測定

抵抗測定

0～500MΩ

0～30kΩ

確度±20％rdg（電 池 駆 動 時 ）0～3kΩ
3k～30kΩ±(0.5％rdg＋0.2Ω)±(0.5％rdg＋2Ω)

分解能

0.1MΩ

0～3kΩ
3k～30kΩ0.1Ω
1Ω

サンプリング速度

約 1s

約 0.5s

印加電圧 2.5V

10μA 定電流方式

測定方式※ 抵抗測定 2 線式の場合、リード線抵抗は含まず

11-4

１１．１

仕様

■ 表示・機能

表示

時刻

表示器

バックライト付き液晶表示

解像度

255×160 ドット

表示内容

測定データ、設定リスト、Y-T モニタ

設定

年、月、日、時、分、秒

確度

日差 ±1 秒 （23℃±5℃）

USB、RS-232C
インターフェース
測定モード

測定点切換方式

測定スタートコントロールの受信、測定データなどの送信、外部プリン
タ・外部表示器への出力

機能

イニシャル、ダイレクト、メジャー、各点設定可 (温度測定はダイレクトのみ)
チャンネル*0 から*4 まで自動切換測定（CSW-5A・B 接続
スキャニング時のみ *はボックスナンバ）、ジャンプ可能
モニタチャンネルの繰返し測定
モニタ
時間変化に対するグラフィックモニタ
スタートキースイッチ、インターバルタイマ、RS-232C、USB
各点ごとに設定可能

プログラム設定

シンプルメジャー係数±(0.0001～99999)

単位με､mV､℃､kN、mm など 40 種類

小数点

小数点以下の表示を 0～6 桁任意に設定可能

初期値

任意測定チャンネルごとに書込み可能

センサモード

接続するセンサ種別を設定可能

係数1.0000単位

センサモードに連動

小数点

センサモードに連動

自己診断機能バージョン表示、バッテリ、ばらつき、バーンアウト

TEDS 機能規格 IEEE1451.4 クラス 2
機能 TEDS センサ情報の読出し
機能

スタート回数

設定した時間間隔、時刻による自動スタート
時間・分・秒、最大 99 時間 59 分 59 秒までステップごとに設定可能
1 ステップあたり最大 99 回または無限回

ステップ数

最大 5 ステップのプログラム可能

実時刻スタート

ステップごとにスタート時刻(日・時・分・秒)を設定可能

GOTO ステップ

以前のステップにプログラムループ可能

スリープ機能測定時間の 5 秒前に電源 ON、測定終了後自動的に電源
OFF になる。 スリープ機能 ON/OFF 設定可

機能

測定データの記録・再生

記録内容

測定モード、チャンネルナンバ、測定データ、時刻デー
タ、データナンバ

データ容量

最大約 80,000 データ

データ保持期間

約 20 日間 (満充電時)

カード規格

CF カードカード容量

最大 2GB

インターバル
インターバルタイマ

データメモリ

メモリカードオートパワーOFF

キー操作、各インターフェースからコマンドを任意設定時間受けないと自動的に電源を OFF する。オートパワーOFF 機能の ON/OFF 設定可

11-5

１１．２ 標準付属品■ 総合仕様耐振性

29.4m/s２ （50Hz 0.6mm p-p）

耐衝撃性49m/s２防滴性

IP-54 相当 （コネクタキャップを装着した状態）

連続使用時間アルカリ乾電池使用：約 10 時間（ひずみ測定 350Ωブリッジ接続時）

使用温湿度範囲-10～＋50℃ 85％RH 以下(結露を除く)

保存温度範囲-20～＋60℃

電源

単 3 形アルカリ乾電池 4 本
専用 AC アダプタ
または、外部電源入力 DC9～18V

外形寸法

102(W)×49(H)×223(D)mm （突起部を除く）

質量

約 800g１１．２ 標準付属品

11-6

取扱説明書

1部

単 3 形アルカリ乾電池

4本

ショルダーベルト

1本

アクセサリボックス

1個

保証書

1部

１１．３

オプション

１１．３ オプション
品名・型名

概要

RS-232C ケーブル
CR-5532

Dsub9P-10P(小型)クロス 1.5m （TC-32K 専用）
パソコンとの接続に使用します

USB ケーブル
CR-6187

miniB-A （フェライトコア付）
パソコンとの接続に使用します

データケーブル
CR-4521

BNC-10P(小型)1.5m （TC-32K 専用）
EDU-11 との接続に使用します

外部プリンタDPU-S245

測定データを印字出力します

プリンタケーブル
CR-4511

8P-10P(小型)ストレート 0.5m （TC-32K 専用）
DPU-S245 との接続に使用しますAC アダプタ
CR-1867

AC100～240V に接続し、電源供給をします

リモートパワーコントローラ
RPC-05A

スリープ機能を制御することにより、外部バッテリ駆動による長期測定が可能ですスイッチボックス
CSW-5A、CSW-5A-05
CSW-5B、CSW-5B-05

測定点数を 5 点に拡張します

挿入型傾斜計用アダプタ
IA-32

2 軸挿入型傾斜計の X 軸、Y 軸の 2 チャンネルを同時にモニタ可能です

1 ゲージ 4 線式専用アダプタ
CR-5810

NDIS コネクタに接続することにより、1 ゲージ 4 線対応モジュラーコネクタ付きのひずみゲージを接続できます

CF カード

最大 2GB（当社指定）11-7

１１．４ 外観図

１１．４ 外観図

11-8

１２章

第

エラーメッセージ
１２.１ エラーメッセージの解説と対処方法 ·················

12 - 2

１２．１ エラーメッセージの解説と対処方法

１２．１ エラーメッセージの解説と対処方法
本器を動作中に異常が発生したり、誤った操作をした時にエラーメッセージを表示します。
それぞれのエラーメッセージについて解説と対処方法を示します。
エラーの分類

エラーメッセージ

解説と対処方法

ファイルエラーバージョンアップファイルが存在しないかファイルが複数あります

CF カードにバージョンアップファイルが存在しない
場合はバージョンアップファイルを CF カードに入れてください。また、バージョンアップファイルが複数ある場合は、バージョンアップファイルのいずれ
か 1 つだけ CF カードに入れ、バージョンアップを行ってください。

ソフトウェアエラー

前回バージョンアップに失敗した為本体を初期化していますバージョンアップ中の電源 OFF などによりバージョンアップが中断したときに再度電源を立ち上げると表示します。初期化しているため、初期化が終わるまでお待ちください。

バージョンアップエラー

バージョンアップファイルのある
CF カードを準備してくださいバージョンアップ中断時の初期化後に表示します。バージョンアップファイルの入った CF カードを挿入してください。バージョンアップファイルの入った CF カードを挿入すると自動でバージョンアップ
を開始します。

ファイルエラーバージョンアップファイルの確認中です。しばらくお待ちください

CF カード内のバージョンアップファイルを確認しています。バージョンアップファイルの確認が終わるまでお待ちください。

バージョンアップに失敗しました
本体を再起動してくださいバージョンアップに失敗したときに表示します。本
体を再起動してください。再起動後、画面の指示に従って操作をしてください。
バージョンアップ時の操作については「８．６ メンテナンス」を参照してください。

ファイルエラーバージョンアップファイルが破損していますバージョンアップファイルが破損しています。必要なデータを取出した後、CF カードをフォーマットし、新たにバージョンアップファイルを入れてバージョンアップをしてください。
それでも改善しない場合は最寄りの営業所にご連絡ください。

データメモリエラー

データメモリにデータがない為表示できません

データメモリが空のときに読出しを行うと表示します。データが保存してあるときに読出しを行ってください。

バージョンアップ
エラー

12-2

１２．１

エラーの分類

エラーメッセージの解説と対処方法

エラーメッセージ

解説と対処方法

データメモリの残量がありません

データメモリの容量が足りないときに表示します。
データメモリ内の不要なデータを削除してデータメ
モリの容量を確保してください。
データメモリの操作については「６．２ データメモ
リ」を参照してください。

ファイルコピｰエラー

同名のファイルが存在する為コピーできません

同名のファイルが存在しているときにコピーを行うと表示します。ファイル名を変更してください。ファ
イル名の変更については「６．３ CF カード」を参照してください。

ファイルコピー
エラー

ファイルのコピーに失敗しました

データメモリのデータを CF カードにコピーするときに失敗すると表示します。
CF カードを認識しているか確認してください。

CF カードが挿入されていません

CF カードへの記録を ON に設定し、CF カードが挿入していないときに表示します。
CF カードを挿入するか、CF カードへの記録を
OFF にしてください。
CF カードへの記録の操作については「６．３ CFカード」を参照してください。

CF カードの残量がありません

CF カードの容量が足りないときに表示します。CF
カード内の不要なデータを削除して CF カードの容量を確保してください。
CF カードの操作については「６．３ CF カード」を参照してください。

データメモリ、CF カードの残量がありません

データメモリ、CF カードの容量が足りないときに表示します。データメモリ、CF カード内の不要なデー
タを削除してデータメモリ、CF カードの容量を確保
してください。
データメモリの操作については「６．２ データメモ
リ」を参照してください。
CF カードの操作については「６．３ CF カード」を参照してください。

設定値を確認してください

設定可能範囲を超えている場合に表示します。
例 インターバル測定のタイマー設定では 1 秒～
99 時間 59 分 59 秒までの設定が可能です。この設定範囲を超える値を入力すると入力値エラーになります。

データメモリエラー

CF カードエラー

CF カードエラー

データメモリ/
CF カードエラー

入力値エラー

12-3

１２．１ エラーメッセージの解説と対処方法

エラーの分類

エラーメッセージ

解説と対処方法

ADC エラー

ADC エラーが発生しました

Ａ/Ｄ変換回路、通信回路の不良が考えられます。最寄りの営業所にご連絡ください。

RS 出力がタイムアウトしました

RS-232C 出力の通信ができていないときに表示します。RS-232C 出力を使用しない場合はデータ出
力先の設定を OFF にしてください。
データ出力先の設定の操作については「７．３ データの出力」を参照してください。
RS-232C 出力を使用する場合は、出力する機器に対応するフロー制御設定に変更してください。
フロー制御の設定については「７．２ RS-232C の設定」を参照してください。

インターバル測定が設定されていません

インターバル測定の設定をしていないときに測定を開始すると表示します。インターバル測定の設定を行ってください。
インターバル測定の設定については「５．８ 自動測定」を参照してください。

次の測定が 30 秒以内にある為電源を切れません

次の自動測定が 30 秒以内のときに電源を OFF しようとすると表示します。電源を OFF する場合は自動測定を停止してください。自動測定の操作については「５．８ 自動測定」を参照してください。

スイッチボックス 未接続

マルチチャンネルモード時にスイッチボックスを認識していない場合に表示します。スイッチボックスの接続を確認してください。接続方法については「９．２ スイッチボックス」を参照してください。
また、スイッチボックスを使用しない場合はシング
ルチャンネルモードに設定してください。チャンネ
ルモードの設定については「４．４ チャンネル設定」を参照してください。

---

ディレクトリが存在しないか
CF カードが壊れています

CF カードが損傷している可能性があります。必要なデータを取出した後、CF カードを一度フォーマッ
トしてください。
それでも改善しない場合は新しい CF カードをご利用ください。

---

イニシャル値測定が禁止されています

イニシャル測定が禁止の設定になっています。
イニシャル測定の設定については「５．１０ 測定
補助設定」を参照してください。

タイムアウト
エラー

---

---

---

12-4

保守・サービス
万一、動作不良などの故障が生じた場合は、下記に記載の営業所までご連絡ください。

■引取り修理・保守について❏お預かりした製品を速く、確実にお納めするために、故障状況、原因と思われる点をお知らせください。
❏製品を直接ご送付いただく場合は、納品された梱包材料か、それに相当する梱包でご送付ください。
❏製品に付属品を必要とする場合は、必ず製品に添え付けてください。

■保証について本製品は厳密な社内検査を経て出荷されておりますが、万一製造上の不備による故障、あるいは輸送中の事故などによる故障の節は、お買い上げいただいた販売店または当社販売員にお申しつけください。
なお、本製品の保証期間は、ご納入日より 12 ヶ月です。この間に発生した故障は当社にて無償で修理いたします。ただし、この期間内でも使用上の取扱いミス、お客様による改造、変更に起因する故障、天災などによる故障・損傷は無償修理から除外させていただきます。
本製品の運用を理由とする損失、逸失利益などの請求、保証につきましては、本書内容の不備や誤り、記載漏れ等にかかわらず、いかなる責任も負いかねますので、あらかじめご了承ください。

本社〒140-8560

東

京

都

品

川

区

南

大

井

6 - 8 - 2

TEL (03)3763-5611

東京営業所〒140-8560

東

京

都

品

川

区

南

大

井

6 - 8 - 2

TEL (03)3763 -5611

札幌営業所〒063-0826

北 海 道 札 幌 市 西 区 発 寒6 条

11-1-1

TEL (011)665 -2600

北関東営業所〒329-0502

栃山

3 3 3 2 - 3

TEL (0285)51-2251

仙台営業所〒981-3133

仙 台 市 泉 区 泉 中 央 1- 9- 2 ア バ ン サ ー ル 泉 中 央 403

TEL (022)725-3378

つくば営業所〒305-0817

茨 城 県 つ く ば 市 研 究 学 園

1 - 2 - 2

TEL (029)868-6705

高崎営業所〒 370-0045

群 馬 県 高 崎 市 東 町 187-2

布 施 ビ ル 1F

TEL (027)345-6631

名古屋営業所〒465-0025

名社

2 - 2 1 0

TEL (052)776 -1781

大阪営業所〒542-0062

大 阪 市 中 央 区 上 本 町 西

5 - 3 - 1 9

TEL (06)6762-9831

明石営業所〒 673-0016

明 石 市 松 の 内 2-4-10

ユ タ カ 第 一 ビ ル 6F

TEL (078)929 -1462

広島営業所〒730-0051

広 島 市 中 区 大 手 町 5-1-1 大 手 町 フ ァ ー ス ト ビ ル 4F-D

TEL (082)207-0663

福岡営業所〒 812-0011

福 岡 市 博 多 区 博 多 駅 前

TEL (092)431 -7205

木

古

県

屋

下

市

野

名

市

下

東

古

区

上

1 - 2 4 - 9

ハンドヘルドデータロガー –TC-32K
2019 年 3 月 12 版

編集

株式会社東京測器研究所

発行

株式会社東京測器研究所■URL http://www.tml.jp/© 2009 Tokyo Measuring Instruments Laboratory Co., Ltd.