Instruction Sheet
Wireless TROLL & Wireless Rugged TROLL Communication Devices
Part Numbers: 0031240 (Wireless TROLL Com); 0031220 (Wireless Rugged TROLL Com)

Description

1Connection status

The Wireless TROLL Com enables wirelesscommunication between a Bluetooth 2.0-enabled Android orWindows device and an instrument deployed on a cable.
You can also use the Wireless TROLL Com as areplacement for a standard wired TROLL Communication
Device.
The Wireless Rugged TROLL Com permits wirelesstransmission between a Bluetooth 2.0-enabled device and aRugged TROLL data logger. You must use a Direct ReadRugged TROLL cable with the Wireless Rugged TROLLCom.

Dust cover for the USB
connection

Red (flashing) = The
communication device,instrument, and Bluetoothenabled device are notconnected.
Red (continuous) = The
communication device andinstrument are connected, but
the communication device is notconnected to the Bluetoothenabled device.
2-ORThe communication device isconnected via USB cable.
Green (flashing) = The
communication device isconnected to the Bluetoothenabled device, but is notconnected to the instrument.
Green (continuous) = The
communication device,instrument, and Bluetoothenabled device are connected.

3On/Off button
Battery charge status:
100% -90%
90% - 75%

475% - 50%
50% - 25%
Less than 25%
5

Cable connector to theinstrument

6Lanyard connector

7

Serial number

Information subject to change without notice. In-Situ, In-Situ logo, Baro Merge, BaroTROLL, HERMIT, HydroVu™, iSitu, Pocket-Situ, RDO,RuggedCable, RuggedReader, SmarTROLL™, TROLL, VuSitu, and Win-Situ are trademarks or registered trademarks of In-Situ Inc. © 2016.
All rights reserved. This product may be covered by patents identified at www.in-situ.com/patents

01/2017 | 1 | 0031242

85. Tap or click the serial number of thecommunication device to pair the device with thephone or computer. The serial number is locatedunder the USB flap.

USB connection to a powersource for charging the internal
battery or wired connection to acomputer

Instruction Sheet

Connecting to VuSitu

ContentslWireless TROLL Coml

USB cablell

1. Open the VuSitu Mobile App. If you havecorrectly paired your Wireless TROLL Com withyour wireless device, and the instrument isavailable, the software will connect and display
readings.

AC wall charger (U.S.A. only, universal charger
sold separately)

If the Searching screen
continues to show, tap"Choose another device" andselect the device you aretrying to connect to.

LanyardCharging the Communication Device
1. Remove the protective cover from the USB slot.
2. Connect the USB cable to the device.

Connecting to Win-Situ 5

3. Plug the USB cable into the wall charger or apowered USB port such as a computer that has acharger plugged in.

1. Open Win-Situ 5 Software.
2. When prompted, "Connect to device now?" clickNo.

4. The device lights will turn on and blink according
to charge level.

3. Plug the USB charging cable into the computer
and Wireless TROLL Com.

A fully-charged communication device will run for up to 4050 continuous hours.

4. Click Preferences, then click Comm Settings.
5. Select the correct Com port used by the charging
cable, then select the communication settings
for the instrument you are connecting. The
following default communication settings aremost common for In-Situ instruments:

Battery Tips
The communication device uses an internal, rechargeable
Lithium-ion (Li-ion) battery to supply power. While thesebatteries last for years with minimal decrease inperformance, there are a few tips to prolong the life of thebattery.
ll

Avoid full discharges and charge the battery moreoften between uses.
Avoid storing the communication device in a hightemperature (above 86° F / 30° C) area.

Connecting to a Wireless TROLL Coml

Baud: 19200

l

Data Bits: 8l

Parity Bits: Evenl

Stop Bits: 1l

Device Address: 1l

Mode: Modbus-RTU

If you cannot connect using these settings, clickthe "Search for Devices" or "Reset All Devices"
button.

A Wireless TROLL Com can be used to connect theinstrument to software if the sonde is deployed on a cable.

6. Click the checkmark, then click the Connect
button in the lower right hand corner.

1. Turn on the Wireless TROLL Com.
2. Make sure the cable is connected to the
instrument as well as the communication device.

Specifications

3. Go to Bluetooth settings on your mobile device orcomputer.

Operatingtemperature-5 to 50° C (23 to 122° F); 95% relative
humidity, non-condensing

4. From the Bluetooth section, search for devices.

800-446-7488

2www.in-situ.com

Storagetemperature-20 to 50° C (-4 to 122° F); 95% relative
humidity, non-condensing

Dimensions

16 x 4.3 x 3 cm (6.3 x 1.7 x 1.2 in.)

Weight165 g (0.36 lb)

Materials

PC / ABS blend, Silicon, Urethane, Stainlesssteel, Brass, Santoprene, Poron®,
Polyethylene, Versapor®, Titanium, PEEK,Viton®

Environmental
rating

IP67

Output options

Bluetooth®, USB

Instruction Sheet

Communication Android®: SPP
protocolWindows®: SPP or USB
Battery type

3.7 V 8600 mAh Lithium rechargeable cell(UBBL19-FL)

Charging
requirements

5 VDC USB charger (1 A or 500 mA)

Warranty

1 yearCertifications

CE, FCC (SSSBC127-X), WEEE

SensorTemperature

Barometric Pressure

Accuracy±2° C max

±3 mbar max

Range-20 to 70° C (-4 to300 to 1100 mbar
158° F)

Resolution

0.1° C

0.01 mbar

Sensor type Fixed

Fixed

Responsetime< 30 seconds

Instantaneous in thermal
equilibrium

Units ofmeasure

Celsius or
Fahrenheit

psi, kPa, bar, mbar, mmHg,
inHg, Torr, atm

MethodEPA 170.1

Piezoresistive

800-446-7488

3

www.in-situ.com