Small Tool Instruments and Data Management

Standard Digital Indicator

ABS Digimatic Indicator ID-C

Catalog No. E4330-543

Digital indicators offer dramatically improved
readability, usability and functionality

Digital indicators that offer dramatically improved ease of

ABS Digimatic Indicator ID-C
The Nakatsugawa plant is in the central industrial park of leafy Nakatsugawa city, in the Gifu
prefecture. This plant specializes in manufacturing sensor products and handles the complete
development-to-production cycle for measuring instruments such as dial gages, test indicators,
Digimatic indicators, and bore gages. It was opened in 1997 as the 12th Mitutoyo plant in Japan.
Using its state-of-the-art production techniques and facilities, this plant continues to provide
Mitutoyo products that are praised around the world and used with confidence.

ual
t
c
A ize
s

ual
t
c
A ize
s
543-390B

543-470B

543-490B

Examples of use
ABS Digimatic indicator ID-C is a standard digital indicator. A large
LCD incorporating 11mm characters (existing products: 8.5mm) is used
to improve visibility, and three large, easy-to-press buttons are usedin the design to make operations easier to perform. In addition, thisaffordable product has various measurement functions, including theability to perform scaling calculations, judge tolerance, hold data, andperform general comparison measurements.

2

ual
t
c
A ize
s

reading measurement values, usability and functionality

1. Large LCD

4. Functions that support measurement

The large LCD incorporates
11mm characters giving 1.5
times the character area ofexisting products (which display
8.5mm characters) making
measurement values mucheasier to read.

The ID-C has various functions,
including the ability to hold data,
Measurement value and tolerance judgment
output data, switch the measuring
direction, judge tolerance, change
the scale factor, and a lock to prevent
misoperation.
Enlarged display of the tolerance judgment result(For details, see page 5.)

11mm

Actual size

2. Three large buttons

5. 330° rotary display 330°

The popular three-large-button design, which is used in products such as the
ABS coolant proof Digimatic indicator ID-N/ID-B, makes buttons easier to press
and operations easier to perform.

ABS measurement mode

INC measurement mode

ABS Digimatic
upright gage

4.267

ABS Digimatic
thickness gage

54.26750 mm
reference
gage

Measured workpiece

54.267

Switches the measurementmode between ABS and INC.
Measured workpiece

Various application products are available that enable you to perform one-touch
quick measurement of the thickness of small parts, papers, felt, lenses, andpipes, as well as the depth of narrow grooves on cylindrical workpieces, groove
depth, and steps.
(For details, see pages 6 and 7.)

Zeroset using a reference gage to perform
a comparison measurement of the difference
between the reference gage and measured
workpiece

Press and hold the SET button(for at least two seconds) to
switch the mode.

Operation in ABS (preset) measurement mode

products for one-touch quick
6. Application
measurement• Switches between the ABS (preset) andINC (zeroset) measurement modes

Preset using a reference gage to measure theabsolute value from a measurement reference

50 mm
referencegage

The display can be rotated 330°,
allowing use at a position where youcan easily read the measurement
value.

• Power switch• Data output (when connected to an
external device)• Data hold (when no external device isconnected)• Parameter setting mode
Count direction switching, tolerance judgment
setting, resolution switching, scale factor
setting, and function lock setting330°

ABS Digimatic
depth gage

Operation in INC (zeroset) measurement mode

7. The ABS (ABSOLUTE) sensor

3. Expanded lifting capability

The original Mitutoyo ABS (ABSOLUTE) sensor, which is capable of relocating
the origin even after turning the power off, enables you to quickly start multipoint measurement. Also, the ABS measurement mode can be restored evenafter measurement in the INC mode, where zerosetting
is possible at any position, improving work efficiency.

The lifting function that moves the spindle up and down has been expanded to
improve work efficiency when using the ID-C mounted on a stand.
For models that have a 12.7mm measuring range,
a lifting lever (special accessory) can be mounted
on the left or right side, improving work efficiency
and smoothness of movement.

8. Inspection certificate provided as standard

A lifting cable (special accessory) provides a maximumof approximately 25.4mm of spindle movement (twicethat of existing models). So for models that have a
12.7mm or 25.4mm measuring range, the spindle
can be moved through the entire measuring range.
(Applied to models that have a 50.8mm measuring
range, the spindle can be retracted by approximately
25.4mm from the extended position.)

Mitutoyo provides an inspection certificate that includes inspection data asstandard to guarantee that every product shipped is of high quality and safe
to use. Upon request, we can also calibrate purchased measuring instruments
and issue a calibration certificate that proves traceability to national (orinternational) standards for a fee. To minimize calibration uncertainty as muchas possible, both the inspection certificate and calibration certificate are issued
after measurement using dedicated testers developed with advanced Mitutoyo
measurement technologies. Note that the inspection certificate cannot be used
to obtain a calibration certificate because the former does not indicate the dateof purchase.

Lifting knob (special accessory)

By using a lifting knob (special accessory) fitted tothe the top of the spindle, you can perform fullstroke operation without directly touching thespindle.

検査成績書

CERTIFICATE OF INSPECTION
Product name : Digimatic Indicator
Graduation : 0.001mm

Lifting lever (finger hook)Standard accessory (only for models that have a measuring range of
25.4mm or 50.8mm)

Certiﬁcate No.1234567890
Model : ID-C112X
Serial No. : 08123456
Temperature : 20°(Standard Temperature)

Code No. : 543-390
Measuring span : 12.7mm

QC Manager : T.Kosuda

Inspection standard : Mitutoyo standard 1)
unit: µm 測定値 error

指示の最大許容差 MPE of indication

3
2
2

広範囲行き精度/Overall
戻り誤差/Hysteresis
繰返精密度/Repeatability

2
1
passedMaximum Permissible error(MPE) of indication

指示誤差 Errors [µm]Based on : ASME B89.1.10M-2001
Traceable to : NMIJ/AIST by JCSS No.0030.
NIST via No.821/276374-08/M9317
+5
○
0 ×

-5

0

○
×

○
×

○
×

○
×

○ : UP○××○× : DOWN

Mitutoyo Corporation

3

×
○

×
○

±3

2
2

+2
1
passed(*) Overall magniﬁcation and linearity
Judgement (判定）: Passed (合格)注) 規格値には量子化誤差±1カウント含まず
The quantizing error (±1 count) is
excluded from the speciﬁcation value

Method of inspection:MITUTOYO (No sign)1)

If dust or coolant gets into the gap between the spindle and main unit while using
the lifting knob, the spindle travel may become rough or the indicator may failaltogether. Therefore avoid using the ID-C in environments containing dust or
coolant mist.

error

unit: µm

A Overall (*)
S Hysteresis
M
E Repeatability

○
×

×
○

1

×
○

×
○

×
○

×
○

×
○

×
○

×
○

×
○

位置 Position(mm)

○
×

×
○

12.7

この検査成績書には安心してご利用いただけますよう出荷時の検査データを記載していますが、校正証明書取得用にはご利用できませんABS Digimatic Indicator ID-C
SPECIFICATIONSMetricISO/JIS Type

Order No.
Resolution 0.001mm/0.01mm
543-390543-390B
543-394543-394B—
543-470B—
543-490B
Resolution 0.01mm
543-400
543-400B
543-404
543-404B—
543-474B—
543-494B

Inch/MetricMeasuring range Accuracy*[1Hysteresis*

Repeatability*

Measuring force

Mass

Remarks

12.7mm
12.7mm
25.4mm
50.8mm

0.003mm
0.003mm
0.003mm
0.005mm

0.002mm
0.002mm
0.002mm
0.002mm

0.002mm
0.002mm
0.002mm
0.002mm

1.5N or less
0.7, 0.6, 0.4N or less
1.8N or less
2.3N or less

170g
170g
190g
260g

—
Low measuring force——

12.7mm
12.7mm
25.4mm
50.8mm0.02mm0.02mm0.02mm0.04mm0.02mm0.02mm0.02mm0.02mm0.01mm0.01mm0.01mm0.01mm0.9N or less
0.5, 0.4, 0.3, 0.2N or less
1.8N or less
2.3N or less

170g
170g
190g
260g

—
Low measuring force——

ISO/JIS Type and ANSI/AGD Type

Order No.
Measuring range Accuracy*[1Resolution .00005"/.0001"/.0005"/0.001mm/0.01mm
543-391543-391B.5"±.00010"/0.003mm
543-392543-392B.5"±.00010"/0.003mm
543-395
543-395B.5"±.00010"/0.003mm
543-396543-396B.5"±.00010"/0.003mm—
543-471B
1"±.00010"/0.003mm—
543-472B
1"±.00010"/0.003mm—
543-491B
2"±.00020"/0.005mm—
543-492B
2"±.00020"/0.005mm
Resolution .0005"/0.01mm
543-401
543-401B.5"±.0010"/0.02mm
543-402543-402B.5"±.0010"/0.02mm
543-405
543-405B.5"±.0010"/0.02mm
543-406543-406B.5"±.0010"/0.02mm—
543-475B
1"±.0010"/0.02mm—
543-476B
1"±.0010"/0.02mm—
543-495B
2"±.0015"/0.04mm—
543-496B
2"±.0015"/0.04mm

Hysteresis*

Repeatability*

Measuring force

Mass

Remarks.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm.00010"/0.002mm

1.5N or less
1.5N or less
0.7, 0.6, 0.4N or less
0.7, 0.6, 0.4N or less
1.8N or less
1.8N or less
2.3N or less
2.3N or less

170g
170g
170g
170g
190g
190g
260g
260g

—
—
Low measuring force
Low measuring force————.0010"/0.02mm.0010"/0.02mm.0010"/0.02mm.0010"/0.02mm.0010"/0.02mm.0010"/0.02mm.0010"/0.02mm.0010"/0.02mm.0005"/0.01mm.0005"/0.01mm.0005"/0.01mm.0005"/0.01mm.0005"/0.01mm.0005"/0.01mm.0005"/0.01mm.0005"/0.01mm

0.9N or less
170g
0.9N or less
170g
0.5, 0.4, 0.3, 0.2N or less 170g
0.5, 0.4, 0.3, 0.2N or less 170g
1.8N or less
190g
1.8N or less
190g
2.3N or less
260g
2.3N or less
260g

—
—
Low measuring force
Low measuring force————: ANS/AGD TypeNote) Products with an Order No. suffixed “B” have a flat back, and other models have a back with a lug.
* Overall hysteresis and repeatability specifications are valid for normal measurement at 20°C, and the quantizing error of ±1 count is excluded.

COMMON SPECIFICATIONS• Display: 6-digit LCD, sign
• Contact point: Spherical tip SR = 1.5mm (carbide tipped),part No. 901312 (for ISO/JIS Type)part No. 21BZB005 (for ANSI/AGD Type)• Spindle orientation for measurement:- Standard model that has a 12.7mm measuring range: No restrictions- Standard model that has a 25.4mm or 50.8mm measuring range: Normallyat any position between the spindle pointing vertically downward to thespindle horizontal. To perform measurement with the spindle pointing above
the horizontal requires a reverse-position coil spring (special accessory).
- Low measuring force models: See 'Setting measuring force on low
measuring force models' on page 4.

• Position detection method: Capacitance type absolute linear encoder• Battery: SR44 (silver oxide button cell) × 1, part No. 938882• Battery life: Approximately 7,000 hours of continuous use• Maximum response speed: Not restricted (except for scanning measurement)• Service temperature range: 0 to 40°C
• Storage temperature range: 0 to 60°C

4

Setting measuring force on low measuring force models•543-404/404B/405/405B/406/406B•543-394/394B/395/395B/396/396B

Maximum
Spindle orientation
Spring Weight(approximately 0.1 N) measuring force
Yes
Yes
0.5N
Yes
No
0.4N
Pointing vertically downward
No
Yes
0.3N
No
No
0.2N
Horizontal
Yes
No
0.2N
Note) Operation using configurations other than shown above is not guaranteed.

WeightMaximum(approximately 0.1 N) measuring force
0.8N (0.3+0.4+0.1=0.8)
Yes
Yes
0.6N
Yes
No
Pointing vertically downward
0.4N
No
Yes
No
No
Not guaranteed
Horizontal
Not guaranteedNote) Operation using configurations other than shown above is not guaranteed.
Spindle orientation

SpringDIMENSIONS
ISO/JIS Type
19.2

20

50.1

28.8

84.8
70

0
ø8 -0.0090ø8 -0.009

19.5
52

25.4mm range models

65.3

M2.5X0.45

7.3

40.8

26

12.7mm range models

M2.5X0.45

7.3

19.5

7.6

110.6
97.3

M2.5X0.45
7.3

11

ø59

0
ø8 -0.009

17
21.2

31.5

ø59

ø54
.5

ø6

54.4
46.5

19.2

11

85.2

5
ø59

31.5

57.7

ø10.9

50.8mm range models

ANSI/AGD Type
ø 10.9
.43"

28.8
1.13"

19.2
.76"

19
3/4"

31.5 11
1.24" .43"

31.5 11
1.24" .43"

19.2
.76"

85.2
3.35"

59
2.32"

ø 54
2.13"

ø 59
2.32"

57.7
2.27"

ø

6.5
ø 1/4"
83.85
3.30"
67.7
2.67"

17.2
.68"

42.15
1.66"
6.35
1/4"

· Contact point
No.4-48 thread

25.4mm range models
52
2.05"

* is the symbol denoting American Gage Design (AGD).
It shows conformance to certain dimensions for Dial Indicators,
as speciﬁed in ASME / AGD 2, intended to promote interchangeability.
Only applicable to models with an "E" sufﬁx.

26
1.02"

12.7mm range models

66.65
2.62"
6.35
1/4"

ø 9.52 -0.03.375"

109.65
4.32"
95
3.74"

0

17.2
.68"

7.6
.3"

ø 59
2.32"

0

ø9.52 -0.03.375"· Contact point
No.4-48 thread

6.35
1/4"

50.8
2"
44.2
1.74"
20 14.7.79" .58"

50.1
1.97"

6.35
1/4"

0

9.52 -0.03.375"· Contact point
No.4-48 thread

50.8mm range models

5Functions• Zero-setting function(INC measurement mode)• Preset function (ABS scale origin setting)• 330° rotary display

330°• Data output(when connected to an external device)330°

The preset value can be changed easily by using
the SET (digit movement) and MODE (valuechange) buttons.

• Resolution switching(For 0.001mm or .00005" resolution models)

From the output terminal, measurement data canbe output to a PC via a compact printer, Digimatic
mini processor DP-1VR, or input tool by pressing
the button below the display. Wireless transmissionof measurement data to a PC can also be performed
using the measurement data wireless communication
system U-WAVE.

Models with 0.001mm resolution are capable ofdisplaying in 0.01mm resolution. Models with.00005" resolution are capable of displaying in.0001" and .0005" resolution. Select the resolution
according to the application.

• Switching the direction

The measuring direction can be reversed.

• Display value holding(when no external device is connected)

Connecting cables for data output (special accessory)
1m: No.905338
No.264-504
2m: No.905409
DP-1VR(special accessory)• Judging the tolerance

Performs judgment (OK, +NG, -NG) according to
the set upper and lower limit values and displays
the result as a symbol. Enlarged display of the OK
and NG symbols is possible.

Measurement value andtolerance judgment• Calculation: f(x) = Ax

Mounting the ID-C on a measuring jig and setting
the calculation factor (to any value) allows direct
measurement without using a conversion table andimproves measurement efficiency.

Enlarged display of thetolerance judgment result

Standard accessories• Operation manual• Inspection certificate
1 Lifting lever (finger hook)(Only for models that have a 25.4mm or 50.8mm measuring range)• Silver oxide button cell for the monitor: SR44 × 1
Part No. 938882• Weight (only for low measuring force models)Special accessories
2 Lifting leverPart No. 21EZA198(for models that have a 12.7mm measuring range ISO/JIS Type)Part No. 21EZA199(for models that have a 12.7mm measuring range ANSI/ADG Type)

Input tool
No.264-012-10 IT-012U (special accessory)• Function locking

Ensures reliability of measurement by locking thesettings to prevent preset function settings frombeing changed by mistake.

• Low battery voltage alarm• Error alarm• Interchangeable contact points for Mitutoyo dial gages

Flat

Shell

Needle		 Various types of contact points are available.

• Various backs for standard Mitutoyo (2 series) dial gages• Reverse-position coil spring
Part No. 02ACA571(for models that have a 25.4mm measuring range)
Part No. 02ACA773(for models that have a 50.8mm measuring range)• Connecting cable (1m), part No. 905338• Connecting cable (2m), part No. 905409

3 Lifting cable, part No. 540774• U-WAVE(measurement data wireless communication system)• Digimatic mini processor DP-1VR No. 264-504• Multiplexer MUX-10F, No. 264-002• Display unit EC-101D, No. 542-007• Input tool (USB keyboard signal conversion type)IT-012U, No. 264-012-10

6

8

2

4 Lifting knob

Part No. 21EZA105(for models that have a 12.7mm measuring range ISO/JIS Type)Part No. 21EZA150(for models that have a 12.7mm measuring range ANSI/ADG Type)Part No. 21EZA197(for models that have a 25.4mm measuring range)
Part No. 21EZA200(for models that have a 50.8mm measuring range)

7
4
5 No.7007

1

3

6• Recommended stands
5 Dial gage stand, No. 7001 7002 7007
6 Granite comparator stand BSG-30, No. 215-154
7 Comparator stand BSC-30, No. 215-504
8 Comparator stand BSG-20, No. 215-151

Application Products
ABS Digimatic Thickness Gage

• For one-touch quick measurement of the thickness of small parts,
papers, and felt
• Rust-free ceramic contact point and anvil (except No. 547-401)
Unit: mm
120
ø10
30

ø10

85

175

547-301547-301547-321

Metric547-321547-401

ISO/JIS Type
Parallelism of contact point
and anvil
ø10mm flat
10µm or less
ø10mm flat
10µm or less
ø6.3mm flat (carbide tipped) 3µm or less

Order No.

Resolution

Measuring range Throat depth Contact point and anvil

Accuracy

Measuring force

547-301547-321547-401

0.01mm0.01mm0.001mm

0~10mm
0~10mm
0~12mm

±20µm
±20µm
± 3µm

1.5N or less
1.5N or less
3.5N or less

Inch/Metric

ANSI/AGD Type

Order No.

Resolution

Measuring range Throat depth Contact point and anvil

Accuracy

Measuring force

547-300S
547-320S
547-400S.0005"/0.01mm.0005"/0.01mm.00005"/0.001mm

0~.4"/0~10mm
30mm
0~.4"/0~10mm 120mm
0~.47"/0~12mm 21mm±.001"±.001"±.0001"

1.5N or less
1.5N or less
3.5N or less

30mm
120mm
21mm

Parallelism of contact point
and anvil
ø10mm flat
.0005" or less
ø10mm flat
.0005" or less
ø6.3mm flat (carbide tipped) .0001" or lessNote) 1. Changing the contact point requires total adjustment, including the main display unit. Contact Mitutoyo for advice.
2. The accuracy specification does not include the quantizing error of ±1 count.
3. Left-handed models can be supplied to special order.

ABS Digimatic Lens Meter

Unit: mm

MetricFlat contact
point

9.3

Contact
point

SR3

Anvil

Concave lens

ISO/JIS Type

9.3

SR2
Anvil
Convex lens

Anvil
Moon lens

Sø4 sphericalcontact point
Standard accessory
Anvil adjustment:
12 mm
SR3

ø6
ø4.8

30

• Designed for measuring the thickness ofconcavo-convex lenses and flat objects• The thickness of flat objects can be measured
by replacing the anvil.
• Includes a spherical contact point.

Spherical contact Flat contact
point
point

30.3

(Throat depth)
85

Resolution

Measuring range Throat depth

Anvil adjustment

Accuracy

Measuring force

547-3130.01mm

0~10mm

12mm±20µm or less

Inch/Metric

ANSI/AGD Type30
1.5N or less
ø6
ø4.8

Order No.

Resolution

Anvil adjustment

Accuracy547-312S.0005"/0.01mm 0~.4"/0~10mm 30mm

12mm±.001" or less

Measuring range Throat depth

SR2

30

30mm

9.3

Order No.

SR3

Measuring force

Anvil displacement: 12 mm
85
1.5N
or lessNote) The accuracy specification does not include the quantizing error of ±1 count. Left-handed models can be supplied to special order.

7

Unit: mm

ABS Digimatic Pipe Gage

ø3

ø3

Contact
point

Tip: ø3.5

Effective
throatdepth• Designed for measuring the thickness of pipes and curved boards

ISO/JIS Type

Order No.

Resolution

Measuring range Throat depth547-3600.01mm

0~10mmInch/Metric

ANSI/AGD Type

Order No.

Resolution

547-361S.0005"/0.01mm 0~.4"/0~10mm 20mm

20mm

Measuring range Throat depth

85

Minimum inside
diameter of pipe
ø3.5mm

Accuracy

Measuring force±20µm or less

1.5N or less

Minimum inside
diameter of pipe
ø3.5mmアンビル

Accuracy

Measuring force±.001" or less

1.5N or less

ø3

Metric

Anvil

20
21

有効差し
込み深さ
Note) 1. Changing the contact point requires total adjustment, including the main display
unit. Contact Mitutoyo for advice.
球ø3.5

21

2. The accuracy specification does not include the quantizing error of ±1 count.
3. Left-handed models can be supplied to special order.

ABS Digimatic Groove Gage

Metric1

Contact
point
Anvil

66

• Best suited for measuring the depth of narrow grooves on
cylindrical workpieces• The contact point and measuring face of the anvil are
1mm thick blade blades.

30.3

ISO/JIS Type

Order No.

Resolution

547-315

Measuring range Throat
depth

0.01mm

0~10mmInch/Metric

ANSI/AGD Type

Order No.

Resolution

547-316S.0005"/0.01mm

Unit: mm(Throat depth)85

 ontact point and anvil Accuracy
C
Measuring force
dimensions (W × D)
6.5mm30x 1mm±20µm or less 1.5N or less
ø6.5

55

30mm

Measuring range Throat
depth 10~4"/0~10mm 30mm

 ontact point and anvil Accuracy
C
Measuring force
dimensions (W × D)
85
6.5mm x 1mm±.001" or less 1.5N or lessNote) 1. Changing the contact point requires total adjustment, including the main display unit. Contact Mitutoyo for advice.
2. The accuracy specification does not include the quantizing error of ±1 count.
3. Left-handed models can be supplied to special order.

ABS Digimatic Depth Gage

Unit: mm

ø59

(67)

(105)

• Suitable for measuring the depth of holes, narrow grooves, and steps.
• The lifting lever can be used either on the left or right sides.

MetricISO/JIS Type

Order No.
547-211547-212547-251547-252

Resolution
0.01mm

Accuracy±20µm

0~200mm 12mm
0.001mm

Inch/Metric

ANSI/AGD Type

Order No.

Resolution

547-217S
547-218S
547-257S
547-258S

Measuring Stroke
range

Measuring
range.0005"/0.01mm 0~8".0005"/0.01mm 0~8".00005"/0.001mm 0~8".00005"/0.001mm 0~8"

±5µm

Base
LengthWidth Flatness
63.5mm
1.5N or less
5µm or less101.6mm
16mm
63.5mm
1.5N or less
2µm or less101.6mm
Measuringforce

Measuringforce±.001" or less 1.5N or less±.001" or less 1.5N or less±.0002" or less 1.5N or less±.0002" or less 1.5N or less

Stroke Accuracy
.5"
.5"
.5"
.5"

Note) The accuracy specification does not include the quantizing error of ±1 count.

8

Base
Length
2.5"
4"
2.5"
4"

Width
.63"
.63"
.63"
.63"

17

(10)

(35)
(11)

ø16

ø5
ø6

SR

1.5

A

Contact point: Carblde- Extension rods
tipped ball
21JAA224

Contact point: Carbldetipped ball
Flatness.0002" or less.0002" or less
21JAZA242.00008" or less.00008" or less

5 pcs.
(10, 20, 30, 30, 100mm)

Extension rods
4 pcs. (.5",1",2",4")
4 pcs. (.5",1",2",4")
4 pcs. (.5",1",2",4")
4 pcs. (.5",1",2",4")

ABS Digimatic Bench Gage
Unit: mm
40

• The support plate allows easy reading by tilting the anvil.
• Carbide measuring faces (for the contact point and anvil)• The lifting lever can be used either on the left or right sides.
ø59

92
60

5

MetricSupport plate

ISO/JIS Type

Order No.
547-064

Indicator stroke
12mm

Inch/Metric
Order No.
547-066S

Resolution
0.01mmWorkpiece height
ø20mm

ø20

55

45

Effective throat depth

ø8

ø6.3

10

Flat anvil

Effective throat depth Anvil adjustment
15mm
13mm

Accuracy±20µm or less

Measuring force
1.5N or less

Resolution
Remarks (contact point) Effective throat depth Workpiece height.0005"/0.01mm .248" DIA flat type
.59"
1"

Accuracy±.001" or less

Measuring force
1.5N or less

ANSI/AGD Type
Indicator stroke.5"Note) The accuracy specification does not include the quantizing error of ±1 count.

Lifting cable• Best suited for inspection of small parts at a site
• Carbide ball contact point• The lifting lever that is a standard accessory for No. 547-054,
547-034S can be used either on the left or right sides.

Anvil(hardenedsteel)

Effective measuring range

Unit: mm

ABS Digimatic Upright Gage

SR

35

12

115

1.5

ø40

MetricInch/Metric
Order No.
547-034S
547-033S
547-035S

547-055

ISO/JIS Type

Order No.
547-054547-053547-055

Resolution
0.01mm0.001mm0.001mmCenter of anvil
1
Measuring depth 45 20
95

Measuring range
12.7mm
12.7mm
12.7mmAnvil
ø40mm (steel type)ø50mm (ceramic type)ø40mm (steel type)Measuring range
.5"
.5"
.5"

Anvil
1.57" DIA steel type1.97" DIA ceramic type1.57" DIA steel type

Workpiece height
30mm

Accuracy±20µm or less
± 3µm or less
± 3µm or lessCenter ofcontact point

60

Measuring force
1.5N or less

ANSI/AGD Type
Resolution.0005"/0.01mm.00005"/0.001mm.00005"/0.001mm

Workpiece height
1.2"
1.2"
1.2"

Accuracy±.001" or less±.0001" or less±.0001" or less

Measuring force
1.5N or lessNote) 1. The lifting lever (part No. 21EZA198) is a standard accessory for 547-054.
2. The lifting lever (part No. 21EZA199) is a standard accessory for 547-034S.
3. The lifting cable (part No. 540774) is a standard accessory for 547-055, 547-033S, 547-035S.
4. The accuracy specification does not include the quantizing error of ±1 count.
• The ABS (ABSOLUTE) scale used in these products is a capacitance type absolute encoder. Its patent has been registered in Japan, the U.S., the U.K., Germany, and China.

9Various Digimatic Indicators

2
1 ID-C: Standard Digital Indicator

52 ID-N/B: waterproof, 35mm slim bodywith various functions

3 ID-H: infrared remote controller, high3

accuracy, and various functions

4 ID-S: cost-effective type that has basic

functions

1

5 ID-U1025: general-purpose type thathas a 25.4mm measuring range

4

6 ID-C112RB: has a built-in calculation

function

7 ID-C112A: has a peak hold function
6

8 ID-C112GB: internal diameter measuring

instrument dedicated to cylinder gages

1

Export permission by the Japanese government may
be required for exporting our products according to
the Foreign Exchange and Foreign Trade Law. Please
consult our sales ofﬁce near you before you export
our products or you offer technical information to anonresident.

Mitutoyo Corporation

20-1, Sakado 1-chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
http://www.mitutoyo.co.jp
Note: All information regarding our products, and in particular the illustrations, drawings, dimensional and performance data contained in this pamphlet, as well as othertechnical data are to be regarded as approximate average values. We therefore reserve the right to make changes to the corresponding designs, dimensions and weights.
The stated standards, similar technical regulations, descriptions and illustrations of the products were valid at the time of printing. Only quotations submitted by ourselves
may be regarded as deﬁnitive.

149 0903 1 C (CH) NE, Printed in Japan8
7