| CONNECTION DIAGRAM

FRONT

1.4F.1

24VDC (OUT)

F.2

GND

F.3

RS-485A+ (RX/TX)

F.4

B- (RX/TX)

F.5

POWER
24 VDC RD+24V A.1

BK

GND A.2

USB HOSTWH

1.2 GND

BU

VIB+BK

1.1

Please verify proper
colours depends onsensors type

SHIELDBN

1.3 TEMP
NC+24V D.1

2.4F

WH

2.2 GND

BU

2.1 VIB+BK

3.4

A

B

3.3 TEMP

BNWH
BU

4.4

ETHERNET

D.2

D

GND D.3

WE MAKE INDUSTRY SMARTERSHIELD

3.2 GND
3.1 VIB+B

PMI

BN
2.3 TEMP

BOTTOM VIEWBKSHIELDBN

4.3 TEMP

WH

4.2 GND

BU

4.1 VIB+BK

VIBRATIONSENSOR
1

VIBRATIONSENSOR
2

PHASE
MARKER

SDMS-VMS-E
Vibration Module
Quick Reference Guide

USB DEVICEE

E

VIBRATIONSENSOR
3RDRED

WH

WHITE

BK

BLACK

BU

BLUE

BN

BROWN

VIBRATIONSENSOR
4

The manufacturer reserves the right to make changes tothe guide. The latest and valid version of this document isthe electronic version available on elmodis.com/downloads

DIN RAIL
22.03.30

ELMODIS Sp. z o.o.

Aleja Pokoju 1elmodis@elmodis.com

www.elmodis.com

31-548 Kraków, Polandtel. +48 531 507 668

The SDMS-VMS-E module is part of the Elmodis system dedicated for vibration monitoringof machines driven by electric motors. The SDMS-VMS-E module is an extension of thesystem to measuring vibration and temperature. The function of the SDMS-VMS-E module
is to measure and process signals from vibration sensors. The measured signals are
sampled at a frequency of 32kHz, which provide high quality data. Rotation marker allows
to perform measurements in reference to a specific angular position of the shaft. The
module’s parameters allow for broadband spectral analysis and dedicated indicators ofthe measured vibration signal. The SDMS-VMS-E has an Ethernet TCP/IP interface used
for local data distribution and sharing informatons to the Elmodis cloud based app.

| MAIN FEATURES−

Support for up to four vibrationsensors−

IEPE (Integrated ElectronicPiezoelectric) sensors support−

Temperature measurementfrom vibration sensors withintegrated temperature sensor−

Sampling of sensor signal
at a frequency of 32kHz−

Ethernet interface−

Support for MQTTand MODBUS TCP/IP protocol−

Power supply 24 VDC−DIN rail mounting

SymbolDescription| OPERATION

Vibration measurement range:±50 [g] (for a sensor withthe sensitivity of 100 mV/g)

Nominal supply current ofvibration sensors:

4.7 mA

Measurement
track according to:

ISO 2954 standard

Temperature
measurement range:

10 ÷ 120 [°C] (for a sensor withthe sensitivity of 10 mV/°C)

Temperature measurement
accuracy:±0.5 °C

Power supply (SELV):

24 VDCPower consumption:< 7.5W

Communication:

10/100 Mbit Ethernet

Operation temperature:-20 ÷ 50 [°C]Sensor circuit shorted /sensor damaged

Storage temperature:-20 ÷ 70 [°C]

Operating humidity range:

5% to 90% without condensation

Machine OK

Maximum altitude:

2000 mMounting method:

TH35 rail (acc. to PN-EN 60715 standard)

Dimensions:46 x 115 x 125 [mm]Weight:0.4 kgStable communication
with the Cloud
No communication withthe Cloud
No Internet accessCollecting vibration
measurements

Before starting installation work, read the installation and commissioning manual of theELMODIS system. Before starting installation
work, it is strictly required to turn off the electrical power supply in the switchgear of the machine at which the Elmodis system is installed.
Not complying with the above requirement
may result in an electric shock dangerous tohumans. During installation, take special care
not to damage existing equipment or interrupt
its operation.

Sensor properly connected,
in stand-by modeNo vibration sensor or open
circuit

Module identification
CASING
BACK-LIGHT

ELMODIS.COM

IEPE (Integrated Electronic Piezoelectric)

Service required - contact
Elmodis customer service

Installation can be performed only byqualified personnel.

| TECHNICAL DATA
Vibration sensor standard:

Failure-free operation of thesystem| SAFETY

The symbol designating selective collection of electronic equipment. It isforbidden to dispose used equipmentwith other waste.

Status

Machine-related alarmIntermittent

Static