200250 ProTIM-C
Datasheet

163663 Rev. AA

Bently Nevada Machinery Condition Monitoring

Description
The 200250 dual-input ProTIM-C (conduit mount) provides 2channels of measurement. All acceleration-to-velocity (A-V)
channels condition the signal from an accelerometer andintegrate it to velocity units. The low frequency A-V channels
are better suited for slower speed applications. A-V withAcceleration Enveloping (AE) channels provide both
integrated velocity units and advanced AE signal
conditioning. Temperature channel types include K-type
thermocouples and 100Ω Platinum (Pt) RTDs with electricalisolation. Each channel is independent and specified when
the ProTIM is ordered.
Table 1: ProTIM Measurement Types and Transducers
Measurement TypeInterfaced
Transducer

Acceleration-to-Velocity (General
Purpose)200150

Low Frequency Acceleration-toVelocity

200155

Acceleration-to-Velocity with AE

200157K-type Thermocouple

2001252-, 3-, or 4-Wire Platinum RTD

Industry standard

Rack Buffered Output (RBO)

Monitor

Process Variable (PV)MonitorPressure

120M1644

Displacement

330101/330103

200250 ProTIM-C
Datasheet

163663 Rev. AA

Specifications
All specifications are at +25 ± 5°C (+77 ± 9°F)
unless otherwise specified. Operation outside
the specified limits will result in false or
inaccurate readings.
Table 1: ProTIM and Transducers Frequency
Responses
DeviceLower Freq

Frequency
Response 1

10 Hz to 1 kHz(600 cpm to 60,000 cpm)
±20% (± 2 dB).

Not OK Range

Open transducer signal,
power, or common is “Not OK”.
Shorted leads are “Not OK”
except for SIG+ shorted to SIGor common shorted to shield.

1 This Frequency response represents the System ProTIM& Transducer. For details on individual device frequency
response refer to ProTIM and Transducers Frequency
Responses

Upper Freq

200250-01

1 Hz

1 KHz

200250-05

1 Hz

1 KHz

200250-06 AV 1 Hz

1 KHz 1

200250-06 AE

1 Hz

500 Hz 1

Measurement 0 to 50 mm/s pk
Range(0 to 2 in/s pk)

200150 XDCR

10 Hz

1 KHz

Over Range

63 mm/s pk (2.5 in/s pk)

200155 XDCR

3 Hz

10 KHz 2

200157 XDCR

10 Hz

10 KHzResolution

0.025 mm/s
(0.001 in/s pk) nominal

Accuracy±15% at 80 Hz

Frequency
Response 1

3 Hz to 1 kHz(180 cpm to 60,000 cpm)
± 10% (±0.9 dB) 2.

Not OK Range

Open transducer signal,
power, or common is “Not OK”.
Shorted leads are “Not OK”
except for SIG+ shorted to SIGor common shorted to shield.

Low Frequency A–V Channels (with200155)

1 2002XX-06 ProTIM AE circuitry allows enveloping input

frequency up to 10 KHz.

2 200155 has a long settling time. Therefore, it should

only be used for low frequency acceleration to velocity
channel types.

200250 ProTIM-C and 200150, 200155,
200157 or 200125 transducer systems
For detailed specification on the transducers,
refer to the individual transducer data sheets.

1 This Frequency response represents the System ProTIM& Transducer. For details on individual device frequency
response refer to ProTIM and Transducers Frequency
Responses

Electrical
A–V (General Purpose) Channels(with 200150)

2 The A-V circuitry attenuates frequencies above 1 kHz.

Use of the 200155 transducer to obtain higher frequency
information will be ineffective.

Measurement 0 to 50 mm/s pk
Range(0 to 2 in/s pk)

A–V w/ AE Channels (with 200157)

Over Range

63 mm/s pk (2.5 in/s pk)

Resolution

0.025 mm/s (0.001 in/s pk)
nominalMeasurement 0 to 50 mm/s pk
Range(0 to 2 in/s pk)

Accuracy±15% at 80 Hz

2/17

Over Range

63 mm/s pk (2.5 in/s pk)

Resolution

0.025 mm/s (0.001 in/s pk)
nominal200250 ProTIM-C
Datasheet

Accuracy

163663 Rev. AA

Temperature Channels±15% at 80 Hz

Frequency Response 1

A-V

10 Hz to 1 kHz (600 cpm to
60,000 cpm) ± 20% (± 2.0 dB) 2.
ProTIM and Transducers
Frequency Responses-18°C to +204°C(0°F to +400°F)

Resolution

0.07°C (0.12°F)

Accuracy

10 Hz to 500 Hz (600 cpm to
30,000 cpm) ± 20%(±2.0 dB) 3

AE

Measurement
Range

K-Type TC

Open transducer signal,
power, or common is “Not OK”.
Not OK Range: Shorted leads are “Not OK”
except for SIG+ shorted to SIGor common shorted to shield.

Maximum temperature ramp
rate: ±0.5°C/min.
RTD±4.45°C (±8°F), RTD lead wireerror not included

OK Range-31°C to +213°C(-25°F to +415°F)

1 This Frequency response represents the System ProTIM& Transducer. For details on individual device frequency
response, ProTIM and Transducers Frequency Responses
2 The A-V circuitry attenuates frequencies above 1 kHz.

Use of the 200157 transducer to obtain higher frequency
information will be ineffective. AE signals up to 10 kHz
are processed at the ProTIM.

Not OK
Condition

3 The 500 Hz filter has a 4-pole attenuation slope. The

enveloped signal will range between 1Hz to 500 Hz.

200150Temperatures outside the OK
Range
Open RTD or thermocouple
wires are "Not OK"

RTD Compensation Coefficient Alpha in
W/W/°C

Table 2: ProTIM and Transducers
Compatibility
Device±8°C (±14°F), including
ProTIM-C, thermocouple &lead wire error, maximum
length of 6 meters.

200155 200157

200250-01

Great

N/A

OK1

200250-05

OK2

Great

OK2

200250-06

OK 3

N/A

Great

European

0.00385

US Industrial

0.00392

SoftwareAt host computer
Compensation

1 The ProTIM does not offer AE capability and will only

accept frequencies up to 1 KHz whereas 200157 will go
up to 10 KHz.
2Lower transducer limit is 10 Hz, whereas 200155 will

operate down to 3 Hz.

3 The ProTIM's AE circuit accepts frequencies up to 10 KHz,

but 200150 operates only up to 1 KHz.

3/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

Rack Buffered Output Channels
AC: 1 Vpp to 8 Vpp full scale
Measurement
DC: 0 to –20 Vdc
Range(See Table 2)

Full Scale

1 Vdc to 3 Vdc Bias fromtransducer

Frequency
Response

Less than 3 KHz

Displacement Channels (with330101/330103)

Table 3: Input Signal and Range
Input Signal

OK Range

Overrange

Input

Used with our 3300 XL 8 mmprobe or 3300 5 mm probe andextension cable (5 metre
system) only.

OutputVoltage
Range-3.0 to 3.0 V (Over Specified
linear range)

Transducer
LinearRange

10 to 50 mils (0.254 to 1.27 mm)

Scale
Factor

120 mV/mil +/-10%

Process Variable Channels

Accuracy
Over Gap
Range+/-1.2 mils @ mid-scale range.

Measurement Current: +4 mA to +20 mA
Range
Voltage: +1 Vdc to 5 Vdc

Frequency
Response

DC to 3 KHz (0 to 180,000 cpm).

Minimum
Target size

15.2 mm (0.6 in) diameter (flattarget)

LOAC_IN

1.6 Vpp

1.92 Vpp

HIAC_IN

8 Vpp

9.6 Vpp

DCGAP_IN

DC: -20 VDC
AC: 1 Vpp

DC: -24 VDC
AC: 5.3 Vpp

Resolution

AC: ±1% of full-scale value at 100
Hz
DC: ±500 mV, absolute accuracy

Frequency
Response

10Hz to 3 KHz (+0 to –5%)

Not OK
Range

Input signal is out of range,
Input signal miswired.

Over Range

Current: <+3 mA or >+22 mA
Voltage: <+0.8 Vdc or >+5.5
Vdc

Resolution±1% of full-scale value Typical

OK Range

Current: Over +3.2 mA
Voltage: Over 0.8 Vdc

Frequency
Response

DC to 3 KHz

Pressure Channels
Measurement
0 to 50 mV
Range
Resolution±8.5% of the transducer fullscale rating (transducer
accuracy not included)

4/17

200250 ProTIM-C
Datasheet

Shaft
Diameter:

Not OK
Range

163663 Rev. AA

Minimum: 50.8 mm (2 in)
Recommended minimum: 76.2
mm (3 in)
Measurements on shaft
diameters smaller than 50 mm(2 in) usually require close
spacing of radial vibration oraxial position transducers withthe potential for theirelectromagnetic emitted fields
to interact with one another(cross-talk), resulting inerroneous readings. Careshould be taken to maintainminimum separation oftransducer tips, generally atleast 40 mm (1.6 in) for axial
position measurements or 74
mm (2.9 in) for radial vibration
measurements. Radial vibrationor position measurements on
shaft diameters smaller than
76.2 mm (3 in) will generally
result in a change in scale
factor. Consult Performance
Specification 159484 foradditional information.

Humidity

Apply DC4 grease on connecter
contacts to improve environmental
performance and prevent corrosion.
Enclosure Type

Housing
Material

Powder-coated Aluminum

Weight

620 g (22 oz) not including
conduit bodySee Dimensional Diagram andDimensions Wiring Connection Details on
page 14.

200151 Transducer Cables
Used to connect the 200150, 200150, and2000157 transducers to the ProTIM-C.
OperatingTemperature

Environmental Limits-20°C to +100°C(-4°F to +212°F).

These cables may be used at lower
temperatures down to –40°C(-40°F), if the cable is not allowed tomove or flex. Flexing these cables attemperatures below –20°C (-4°F)
may damage them.

Standard:-40°C to +85°C(-40°F to +185°F)

RTD, K-Type TC and Thermocouple
ProTIM have a limited operating
temperature of -25° C to +85° C.

Storage Temperature

Type 4

Mechanical

Open/short transducer signal,
Power or common is “Not OK”.

OperatingTemperature

100% condensing on
exposed surfaces.
100% noncondensing
on surface inside
conduit.

-40°C to +100°C(-40°F to +212°F)

5/17

Minimum
Bend Radius

63.5 mm (2.5 in)Construction

4-conductor (22 AWG) with
foil shield and drain wire
(100% coverage), polyvinyl
chloride (PVC) outer jacket.

200250 ProTIM-C
Datasheet

Connectors

Classifications:

163663 Rev. AA

Voltage rating 300 Vrms

Screw-on, 5-pin, keyed
connector on the ProTIM-C
end and a PT06F8-4S (orequivalent) on thetransducer end. Connector
coupling nuts consist of 1/220 UNF-threaded 2011 T3
aluminum or UV-stabilized
black nylon. Contact
material is gold-plated
nickel-coated brass.

Capacitance

Cable assembly meets UL
2238.
Cable meets IP67 ingress
protection.

85033 Trendmaster SPA/TIM line
cable
Use to connect a SPA to the ProTIM-C.
For substitutions, reference guide 101206.
OperatingTemperature-70°C to +200°C(-94°F to +392°F).

Conductors

4x 18 AWG stranded tinned
copper
1x 18 AWG stranded copper,
tinned overcoat uninsulated
drain wire

Shielding

100% aluminum mylar foil out
with helically applied drain
wire
85% braided tinned copper

Insulation
Conductors

Fluoroethylene propylene(FEP) Teflon insulation 0.25
mm (0.010 in) thickOuter

FEP Teflon insulation 0.38 mm(0.015 in) thickClassifications

NEC article 725 class 3
UL Listed

6/17

BetweenConductors

131 pF/m (40 pF/ft)

BetweenConductor
and Drain
Wire

262 pF/m (80 pF/ft)

200250 ProTIM-C
Datasheet

163663 Rev. AA

Compliance andCertifications
FCC
This device complies with part 15 of the
FCC Rules. Operation is subject to thefollowing two conditions:llThis device may not cause harmfulinterference.
This device must accept any
interference received, includinginterference that may cause
undesired operation.

EMC
EN 61000-6-2 :2005
EN 61000-6-4:2007 +A1:2011
EMC Directive 2014/30/EU

RoHS
RoHS Directive 2011/65/EU

ATEX
EN 60079-0: 2012/A11:2013
EN 60079-11:2012
EN 60079-15:2010
EN 60079-28:2015 (DSM 149744 only)
EN 60079-31:2014 (TMGI 162459-01 only)
ATEX Directive 2014/34/EU

Maritime
ABS 2009 Steel Vessels Rules
1-1-4/7.7,4-8-3/1.11.1,4-9-7/13

7/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

II (1) G [Ex ia Ga] IIC

Hazardous Area Approvals
TMGI(162459-01)

For the detailed listing of country andproduct specific approvals, refer tothe Approvals Quick Reference Guide(108M1756) available from Bently.com.

T4 @ Ta = -20°C to +65°C

CSA/NRTL/C (Approval Options 05)
Ex ia IIC T4 Ga
Installed withintrinsicallysafe zener
barriers per
drawing
112M7732

Installed
without
barriers per
drawing
112M7732

Class I Zone 0: AEx ia IIC T4 Ga
Class I, Div 1 Groups A, B, C & D
Class II, Groups E, F & GClass III
T4 @ -40° C[ Ta [+ 100° C(-40° F [ Ta [+212° F)
Ex nA IIC T4 Gc
Class I Zone 2: AEx nA IIC T4
GcClass I, Div 2 Groups A, B, C
&D
T4 @ -40° C [ Ta[+ 100° C(-40° F[Ta [+212° F)

ATEX/IECEx
ProTIMs(200200 and200250)II(1) D [Ex ia Da] IIIC
II 3 (1) G Ex nA [is Ga] IIC T4 Gc
II 3(1) D Ex tc [ia Da] IIIC T140°c
Dc IP5X

II 1 G Ex ia IIC T4 GC
II 1 G Ex ia IIC T4 GC
II 3 G Ex nA IIC T4 Gc
T4 @ Ta = -40°C to +100°C
II 3 G Ex nA IIC T4 GC

Trendmaster II 3(3) G Ex nA [ic] IIC T4 Gc
DSM (149744) II 3(3) G Ex nA op is [op is T4 Gc]
IIC T4 Gc
T4 @ Ta = -20°C to +65°C

8/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

Ordering Information
For the detailed listing of country andproduct specific approvals, refer tothe Approvals Quick Reference Guide(108M1756) available from Bently.com.

08

Process Variable

09

Pressure (120M1644)

10

Keyphasor (330101/330103)

11

Displacement (330101/330103)

BB option availability is dependent on AA
option chosen. Not all BB options areavailable with each AA option.

ProTIM-C

C: Approvals

200250-AA-BB-CC-DD

01

None

A: Channel A Input Option

05

Multiple Approvals

01

Acceleration to Velocity (200150)

D: Conduit Body Style

02

K-Type Thermocouple (200125) 1

00

No conduit body03

2 or 3 Wire Pt. RTD

01

Appleton Style C body,malleable iron04

4 Wire Pt. RTD

0205

Low Freq Accel-to-Velocity(200155)

Appleton Style E body, malleable
iron

03

Appleton Style C body,aluminum04

Appleton Style E body,aluminum05

Weatherproof housing mount06

Accel to Velocity w/ AE (200157)

07

Rack buffered Output

08

Process Variable

09

Pressure (120M1644)

10

Keyphasor (330101/330103)

11

Displacement (330101/330103)

1 The 200125 is the recommended Trendmaster sensor

for temperature measurements when the range of a Ktype thermocouple is adequate. We do not recommend
the use of other K-type thermocouples due to theunique electrical isolation requirements of theTrendmaster system, and we highly recommend theuse of only non-grounded RTDs and non-grounded tip
thermocouples to prevent ground loops. Failure tocomply may result in Not OK or NO DATA conditions,inaccurate readings, or ProTIM-C damage. Consult
Bently Nevada for further information.

B: Channel B Input Option
01

Acceleration to Velocity (200150)

02

K-Type Thermocouple (200125) 1

03

2 or 3 Wire Pt. RTD

04

4 Wire Pt. RTD

05

Low Freq Accel-to-Velocity(200155)

06

Accel to Velocity w/ AE (200157)

07

Rack buffered Output

9/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

Transducer Cable (for use with
200150, 200150, and 200157
accelerometers)

4.0 metre (13.1 feet) cable60

6.0 metre (19.7 feet) cableStandard cable, unarmored.

03

Stainless steel over braid(armored) cable0050

0 to 50 PSI SG

0500

0 to 500 PSI SG

2000

0 to 2000 PSI SG

Accessories

B: Armor Option:
02

9.0 meter (29.53 feet) cable

A: Pressure rating

A: Cable Length:4009120M1644-AAAA

Use the 200151 with the 200250 ProTIM onlywith applications using either a 142485
Housing Cable Adapter or a 141887 Conduit
Cable Adapter

2.0 metre (6.6 feet) cable

6.0 meter(19.69 feet) cable

Pressure Transducer

200151-AA-BB-CC

200601620085

Extra Terminal Plugs. For SPA line
connection. 3 terminal plugs
provided with each ProTIM-C
module149326-01

The 200151 Accel Cable
Environmental Boot Field
Installation Kit. Contains
everything needed to install theboot in the field. It includes 10
sets of boots and clamps,
silicone lubricant, easy-to-use
installation tool, and an
instruction sheet. You canpurchase additional boots andclamps separately (see partnumber below). The kit is notcompatible with armored cables
and only compatible with older,
plastic versions ofaccelerometers; not compatible
with existing stainless steel
accelerometers.

04500006

Dow Corning 4, ElectricalInsulating Compound (5.3 Oz).

03814231

Compression Fitting. For 1-inch
conduit body hubs. Seals forcables when installing ProTIM-C
modules without conduit.

85033-0200

300 Meter (1000 ft) Cable. For TIM
line.

C: Nut Option:
00

Standard aluminum couplingnut

02

Nylon coupling nut

10

Knurled aluminum coupling nut

C01, environmental boot option, is no longer
available and is not necessary for use with all
three existing stainless steel accelerometers,
but may be used with previous plastic 200150
sensors.

Transducer Cable
120M1648-AA for use with 120M1644
Pressure Transducer.
A: Cable Length:02

2.0 meter (6.56 feet) cable04

4.0 meter (13.12 feet) cable

10/17

200250 ProTIM-C
Datasheet

85033-0100

150 Meter (500 ft) Cable. For TIM
line

162454

Dust Cap. For temperaturechannel connectors.

04400000

Dust Cap. For accelerometerchannel connectors.

141888Hardware Mounting Kit. For
mounting ProTIM-C modules toweather-proof housings

03810116

Red Silicone Rubber Gasket. Use
between ProTIM-C andweatherproof housing.

163996-01

ProTIM-C Conduit Gasket/Cable
Seal Installation Kit. Consists of a
black silicone rubber gasket/seal,
three nylon hose clamps,
electrical insulating compound,and instructions for installation.

162438-01

Thermocouple and RTD
Connector. With smaller cable
crimp seal.

02180005

163663 Rev. AA

Installation
In order to accommodate the wide range ofcustomer installation needs, a selection ofmounting hardware is available to allow avariety of installation options. In addition,
replacement parts are available for mosthardware components. The following table lists
accessories for use with ProTIM-C modules,
Transducers, and some Trendmasterinstallation hardware.
Table 1: Installation Hardware

Jumpers. For the ProTIM-C
Thermocouple or RTD terminals.

For more information on this product, pleaserefer to:Trendmaster ProTIM-C User Guide (document
161934) and 149744 Trendmaster Dynamic
Scanning Module Datasheet (document
149831).

11/17

200150-AA(See transducer
datasheet 164986
for mounting
options)

General-purpose, casemounted seismictransducer designed
specifically for use with
the Trendmaster systems.

200157-AA(See transducer
datasheet 164986
for mounting
options)

High frequency, casemounted seismictransducer designed
specifically for use withTrendmaster systems.

200151-AA-BB-CC(see 200200
ProTIM-R
datasheet
163662)

Accelerometer Extension
Cable; Two Connectors
Armor is stainless steel
braid.

40113-02

3300XL Connector
Protector Kit. Protection
for 3300XL probe
connectors to minimize
corrosion and otherenvironmental damage.

200152-AA-BB(see 200200
ProTIM-R
datasheet
163662)

Accelerometer Extension
Cable; One Connector

English Thread:
330101
Metric Thread:
330103(See transducer
datasheet for
configuration
options)

3300XL 8 mm Proximity
Transducer System.
Applicable ProTIM devices
use 3300XL transducers
with 5 meter total system
length.

200250 ProTIM-C
Datasheet

163663 Rev. AA

85717-01

Standard (non-I.S.)
Weatherproof Housing.
Polyester W/P Housing, Div2, Type 4X, 4 ProTIM-R Max

00500128

Extra terminal plugs fortransducer connection to
ProTIM-R module. Each
ProTIM-R is provided with 2
terminal plugs

02810005

Jumpers for the 2 awire
RTD ProTIM-R channels

03814237

Hub. 12 mm (1/2 in.) rigid
conduit hub for
intrinsically safe
installations

03817040

Bushing, ground, 0.2 inch
for

Trendmster TIM cable. 150
m (500 FT)

03839129

Conduit Fitting. 12 mm(182 in.) NPT

85033-02-00

Trendmster TIM cable. 300
m (1000 ft)

03839240

Cable Seal. 1/4-in NPT, usewith TIM line cable

88312-01

I.S. Housings (Type 4)
Fiberglass Polyester. I.S.
Polyester W/P Housing, Div
1, Type 4X, 4 ProTIM-R Max

03839241

Cable Seal. 1/4 -in NPT, usewith Velocity, RTD,
Thermocouple

03839242

Cable Seal. 1/4 -in NPT, usewith Displacement Probe
Cable

03839243

Cable Seal. 1/2 -in NPT, usewith Armored Probe cable

03830490

O-ring. 1/2-in Square Cut
for ½ -in NPT fittings

03880243

Thread Seal. ¼ -in for ¼ NPT
fittings

04500006

Electrical Grease. Dow
Corning 4, ElectricalInsulating Compound(5.3 Oz)

136806-01

Conduit T Tim (Allows
branching of TIM line in
conduit)

141887-01

Conduit Cable Adapter;
Single141887-02

Conduit Cable Adapter;
Double142485-01

Housing Cable Adapter

164045

Female TIM Line
Connector

164046

Male TIM Line Connector

164532

TIM Line T Connector(Branches the TIM Line
with the ProTIM-R; fits in
the Female TIM Lineconnector slot)

85033-01-00

88313-01

88314-01

I.S. Housings (Type 4)
Fiberglass Polyester. I.S.
Polyester W/P Housing, Div
1, Type 4X, 2 ProTIM-R Max
I.S. Stainless Steel (SST)
Housing. I.S. SST W/P
Housing, Div1, Type 4X, 4
TIMs Max

88315-01

I.S. Stainless Steel (SST). I.S.
SST W/P Housing, Div1,Type 4X, 2 TIMs Max

85716-01

Standard (non-I.S.)
Weatherproof Housing.
Polyester W/P Housing, Div2, Type 4X, 2 ProTIM-R Max

12/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

Graphs and Figures
Table 1: Channel Types Cross-compatibility

13/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

Note: All dimensions in millimetres (inches) except as noted.

1.
2.
3.
4.
5.

Top view
Front view
Bottom view
Powder-coated aluminum housing
Allow 127 mm (5 in) for total height with connector andcable bend

Figure 1: Dimensional Diagram and Wiring Connection Details

14/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

1. Style C conduit body
2. Style E conduit body
Figure 2: Conduit Body Styles Provided with the ProTIM-C

1.
2.
3.
4.
5.
6.

Style C conduit body
Transducer cables
Transducers
ProTIM-C module
Rigid conduit
SPA line

Figure 3: Installed ProTIM-C in Style C Conduit Body

15/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

1.
2.
3.
4.

Retention nuts (2 places)
Reinforcing plate
GasketproTIM-C module

Figure 4: Installed ProTIM-C Module with Weatherproof Housing

16/17

200250 ProTIM-C
Datasheet

163663 Rev. AA

Copyright 2021 Baker Hughes Company. All rights reserved.

Bently Nevada, Keyphasor, Trendmaster and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes
Business, in the United States and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company.
All other product and company names are trademarks of their respective holders. Use of the trademarks does not
imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

17/17