TEST REPORT: 7191236532-CHM20-01B-RC
Date:

04 MAY 2020

Tel: +65 68851345 Fax: +65 67732912

Client’s Ref:

Email: Randy.CHIN@tuv-sud-psb.sg

Note: This report is issued subject to the Testing and Certification Regulations of the TÜV SÜD Group and the General
Terms and Conditions of Business of TÜV SÜD PSB Pte Ltd. In addition, this report is governed by the terms set out
within this report.

SUBJECT
Bacterial Filtration Efficiency (BFE)
CLIENT
JSB TECH PTE LTD
100 Pasir Panjang Road
#06-02 100 Pasir Panjang
118518 Singapore
Singapore
Attn : Ms Paisley Ang
SAMPLE SUBMISSION DATE / TEST DATE
22 Apr 2020 / 29 Apr 2020
DESCRIPTION OF SAMPLE
One sample labelled as “eGee-Pro ® Mask” was submitted by the above company.

METHOD OF TESTASTM F2101-19 Standard Test Method for Evaluating the Bacterial Filtration Efficiency (BFE) of Medical
Face Mask Materials, Using a Biological Aerosol of Staphylococcus aureus.
Flowrate: 28.3 ± 0.3 L/min
Mean particle size of the challenge aerosol: 3 µm ± 0.3 µm
Test area: Approximately 50 cm2Laboratory:
TÜV SÜD PSB Pte. Ltd.
No.1 Science Park Drive
Singapore 118221

Phone : +65-6885 1333
Fax : +65-6776 8670
E-mail: enquiries@tuv-sud-psb.sg
www.tuv-sud-psb.sg
Co. Reg : 199002667R

Regional Head Office:TÜV SÜD Asia Pacific Pte. Ltd.
1 Science Park Drive, #02-01
Singapore 118221

Page 1 of 4

TEST REPORT: 7191236532-CHM20-01B-RC
04 MAY 2020

RESULTS
Sample Description :eGee-Pro ® Mask

Test sample/controls

Stage 1,
CFU

Stage 2,
CFU

Stage 3,
CFU

Stage 4,
CFU

Stage 5,
CFU

Stage 6,CFU

Sum of Total
plate count forthe 6 sieves,
CFU

AverageCount forControls,CFU-ve Control

0000000+ve Control 1

2554736592581703672182+ve Control 22135377785112452542538

Sample 1000001199.96

Sample 2000002299.92

Sample 3000001199.96

Sample 4000001199.96

Sample 5000001199.96

BFE (%)

Average
BFE (%)

2360

99.95

Page 2 of 4

TEST REPORT: 7191236532-CHM20-01B-RC
04 MAY 2020

Notes :CFU
BFE::

Colony Forming Unit
Bacterial Filtration Efficiency

Remarks :The above test results relate to the samples as received.

MS AW HWEE YINGHIGHER TECHNICAL EXECUTIVEMR RANDY CHIN KOK FEIPRODUCT MANAGERMICROBIOLOGYCHEMICAL & MATERIALS

Page 3 of 4

TEST REPORT: 7191236532-CHM20-01B-RC
04 MAY 2020

Please note that this Report is issued under the following terms :1.

This report applies to the sample of the specific product/equipment given at the time of its testing/calibration. The results are not used to indicate
or imply that they are applicable to other similar items. In addition, such results must not be used to indicate or imply that TÜV SÜD PSBapproves, recommends or endorses the manufacturer, supplier or user of such product/equipment, or that TÜV SÜD PSB in any way“guarantees” the later performance of the product/equipment. Unless otherwise stated in this report, no tests were conducted to determine longterm effects of using the specific product/equipment.

2.

The sample/s mentioned in this report is/are submitted/supplied/manufactured by the Client. TÜV SÜD PSB therefore assumes no responsibility
for the accuracy of information on the brand name, model number, origin of manufacture, consignment or any information supplied.

3.

Nothing in this report shall be interpreted to mean that TÜV SÜD PSB has verified or ascertained any endorsement or marks from any othertesting authority or bodies that may be found on that sample.

4.

This report shall not be reproduced wholly or in parts and no reference shall be made by the Client to TÜV SÜD PSB or to the report or results
furnished by TÜV SÜD PSB in any advertisements or sales promotion.

5.

Unless otherwise stated, the tests were carried out in TÜV SÜD PSB Pte Ltd, No.1 Science Park Drive Singapore 118221.

July 2011

Page 4 of 4