SCOUT220-IS and SCOUT240-IS Portable
Data Collectors
Datasheet

Bently Nevada Machinery Condition Monitoring

109M0226 Rev. AA

Description
SCOUT220IS and SCOUT240IS are intrinsically safe portable data
analyzer solutions designed to monitor machinery conditions inhazardous environments. Both are rated ATEX Zone 1 / Zone 21(for gases and non-conductive dust) and CSA Class 1 / Div 1.
These small form-factor, rugged devices take vibration
recordings and upload the collected data for detailed analysis in
System 1.
This is a two-part solution. The SCOUT data acquisition module
uses Bluetooth to communicate with a wireless, industrial
handheld device running the S1 Collector app.
Vibration recordings are viewed and stored on the handheld
device. We offer these devices with hazardous area ratings up to
Zone 1/21 & Div 1. The handheld runs the Android OS, so you canalso use it like any smartphone or tablet to take pictures andvideo, send text messages, check email, make phone calls, track
your location, create SOS alerts, and run other enterprise apps.
The high performance data acquisition module provides long
battery life, yet is light & rugged. It hangs out-of-the-way on ashoulder strap, allowing you to focus on your job and movesafely between machines. The SCOUT220IS can sample two
sensor channels simultaneously, while the SCOUT240IS supports
up to four channels. Both also support tachometer or Keyphasor
triggering. These data acquisition modules provides up to ten
hours of continuous use, while supporting an Fmax up to 80 kHz
and 12,800 lines of spectral resolution. Both models are rated
IECEx/ATEX Zone 0 / 20 (for gases and non-conductive dust) andCSA Class 1 / Div 1.
The System 1 software, paired with SCOUT220IS and
SCOUT240IS, enables you to track all of your machinery using asingle enterprise monitoring system. Use System 1 to prepare
routes, analyze data, detect long-term trends, and set anddetect alarms. Then transfer the routes between System 1 andyour SCOUT device using WiFi, file transfer, cellular network, or a
USB cable.

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Hardware Features
l

l
l

l

l

l

l

l

l
l

l

l

Battery life of 10 hours allows the datacollector to run all day.
Five year warranty.
Rugged, water, and dust resistant (IP-65),
drop-tested, with intrinsically safe
variants.
Easy to carry: lightweight handheld unit,
shoulder strap for data collector, keepers
for sensors.
Safety breakaways on both shoulder
strap and sensor cables.
Upgradable industrial handheld
smartphone or tablet.
Handheld Push-to-Talk(PTT) button is
configured to start measurements(where available).
Up to four channel simultaneousrecordings (including on-route).
SCOUT240 supports Triaxial sensors.
Up to 12,800 lines of spectral resolution,
80 kHz Fmax.
Supports accelerometer, velocity,
displacement, and voltage-output
sensors of AC and DC coupling types.

lCross Channel FRF measurement &export UFF.
Set machine speed from spectral peak.

Software Features
l

l
l
l

l

l

l
l

l

l
l

Seamless on-and off-route datacollection.
Route overview & easy route navigation.
Fast sensor settling and data collection.
Patented 6Pack time-saving
measurement type.
Synchronous sampling, peak demod, andbump test.
Coast-down, run-up, and interval
capture support.
Custom note and numeric data entry.
Rotor Balancing, dual plane, reports
transferred to System 1.
Data quality warnings and easy-tounderstand alarm annunciation.
Supports operator driven reliability (ODR).
Cross Channel Phase measurement &reports.

The SCOUT portable connects via Bluetooth tothe handheld, which you can then sync with
System 1.

2/16

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Tachometer

Data Collector Specifications
Sensors

Optional IS
Zone 0 rated
laser Tach

Battery-powered, handheld
laser sensor.

Sensor input
Simultaneous
sampling

SCOUT220IS : 2 channels.
SCOUT240IS: 4 channels.

Triggers when beam onreflective tape range 10 cm to2 m.
Speed 6 to 250 krpm.

Compatible
sensor types

Accelerometer, velocity,
displacement, and voltage.

Laser sensor powered byPortable Data Collectors.

AC coupledrange

16 V peak-peak.
Allows for ± 8 V sensor
output swing (± 80 g).

DC coupled
ranges0 to 20 V, -10 to 10 V, -20 to 0
V

Connectors

2 x LEMO connectors for CH1& CH2+
1 X LEMO connector for
Tach/Keyphasor
SCOUT240IS includes a CH
2.4 breakout cable.

Analog todigitalconversion
Sensorexcitation
current

Sensordetection

Optional
Zone 2 rated
laser Tach

Triggers when beam onreflective tape range 10 cm to
2 m. Speed 6 to 250 krpm.

Other sensor
types
supported

Contact, TTL Pulse, Keyphasor

Power supply
to sensor

5 V, 40 mA
Rated for Zone 2, 22 andDivision 2 only.

Input
100 KΩ
Impedance &± 28 V max
max values

24-bit ADC
0 mA or 2.2 mA(configurable), 24 V
maximum
2.2 mA powers IEPE/ICP-type
sensors
Measures sensor bias
voltage.
Warns if short circuit or notconnected.

3/16

Waveform
display± 20 V range

Threshold± 20 V range ± 0.4 V accuracy

Autothresholdmode

Signal ≥ 3 V pk-pk
Machine speed ≥ 60 rpm (1 Hz)
Settles within < 15 sHysteresis

0.2 V to 2.5 V user configurable

Speed range

Minimum: 3 rpm (0.05 Hz)
Maximum: 1,200,000 rpm (20kHz)

Signal pulse
width

10 µs minimum 2% duty cycle(for auto-threshold)Accuracy± (0.1 % + 1 cpm) for speeds up
to 5 kHz (300 kcpm).
± (0.3 %) for speeds above 5
kHz (300 kcpm).

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Parameter Indication
Dynamic
signal range> 95 dB typical at 400 line
resolution

Harmonic
distortion

Less than -70 dB typical.
Other distortions and noise
are lower.

Unitsg or m/s2
in/s or mm/s
mil or mm or µm
peak, or peak-peak, or RMS

Magnitude &cursors

Overall value: RMS, derived pk,
derived pk-pk Dual cursors
Digital readouts on chart

Baseaccuracy

Fmin possible
range

0 to Fmax.
Instrument zeroes all spectral
lines below Fmin.

Resolution(Spectrallines)

400; 800; 1,600; 3,200; 6,400;
12,800 lines
Channels

220
max lines

240
max lines

1

6,400

12,800

2

3,200

6,400

3 or 4N/A

3,200

Frequency
scale

Hz, CPM, Orders.
Linear scale with zooming.

For AC signal: ± 1% of readings
For DC level: ± 1% of full scale(1% is approximately 0.1 dB)

Amplitude
scale

Acceleration, velocity,
displacement.
Linear auto scaling.

Highfrequency
attenuation≤ 0.1 dB > 100 Hz to 10 kHz≤ 3 dB > 10 kHz to 40 kHz
Attenuation tolerances are in
addition to base accuracy.

Window
shapesHanning, Rectangular

Overlap

AC coupling
attenuation≤ 0.1 dB 10 Hz to < 100 Hz≤ 3 dB 1 Hz to < 10Hz

(0, 12.5, 25, 37.5, 50, 62.5, 75,87.5) % Dependent on Fmax
and number of lines
1, 2, 4, 8, 16, 32, 64, 128
Increases sampling time
proportionallyAttenuation
due toIntegration≤ 0.1 dB 10 Hz to < 100 Hz≤ 1.5 dB 1 Hz to < 10 Hz.
Values apply to single
integration (acceleration tovelocity). Double the values fordouble integration(acceleration todisplacement).

Number of
averages
Averagingtypes

Linear, exponential, peak hold

Demodulation
bandwidths

23 bandwidth options from125 Hz to 1,250 Hz and up to 16
kHz to 20 kHz.
Peak demod: 500Hz-40kHz, 140kHz, 2-40kHz, 5-40kHz

6Pack

Up to 40 kHz & 3,200 lines (1channel).
Up to 20 kHz & 1,600 lines (2-3channels).

Spectrum Display
Fmax ranges

25; 50; 100; 125; 150; 200; 300;
400; 500; 600; 800; 1,000;
1,200; 1,600; 2,000; 2,500;
3,000; 4,000; 5,000; 6,000;
8,000; 10,000; 15,000; 20,000;
30,000; 40,000; 60,000;
80,000 Hz or equivalen. CPM
values or orders-based from
1X to 999X.

Spectrum and waveform foreach of low-frequency, highfrequency, demodulation

4/16

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Waveform Display
Charger type

Number of 1,024; 2,048; 4,096; 8,192; 16,384;
samples
32,768Channels

Time
scale

220
max
samples240
max
samples

1

16,38432,7682

8,192

16,3843 or 4

N/A

8,192

Charge rate

Size5.1” W x 5.1” L x 1.4” H

Weight

1.5 lb ≈ 680g

Environmental Limits
2 planes, 2 sensors

Speed Range

3 to 12,000 RPM

Measurement
Type

Acceleration, Velocity,
Displacement

Weight modes

Angle 0° to 360°
Fixed Positions

Remove Trial
Weights

Yes/NoAutomatic recalculation

Storage

Balance jobs stored on
handheld and transferable
to System 1

Operatingtemperature
Ambient
temperature
for charging
battery
Storagetemperature& humidity

Communication

Ruggedness
drop &
sealing

BT 4.0 Classic (s/n ≤ 101599)
BT 5.0 BLE (s/n ≥ 101600)

Battery and Charger

OperatingTime

2 A nominal 3 hours forcomplete charge.

(130 mm W x 130 mm L x 35
mm H

10 ms to 256 seconds or orders
based from 1 to 999 revs

PlanesBattery type

External power pack 12 V DC, 3
A output.

Mechanical

BalancingBluetoothInternal charging, automatic
control.

Custom Lithium Ion pack, 7.4 V,
3920 mAh.
Not user accessible but
factory replaceable.
10 hours typical on-route datacollection

5/16-4°F to 140°F (-20 to 60°C)
122°F (50°C) max when usedin Ex hazardous areas
32°F to 77°F (0°C to 25°C)-4°F to 140°F (-20°C to 60°C),
95%relative humidity If
storage exceeds one month:
up to 95°F (35°C), 85% relative
humidity.
Drop spec MIL-STD-810F516.5-IV 26 drops at 4' (1.2 m)
onto concrete
IP65

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

S1 Collector App
Operating
SystemLanguages

Android 5.1 to 12
Chinese, English, Finnish,
German, Japanese, Russian,
Spanish, Turkish

iOS not approved for use with
SCOUT200 data collectors.

6/16

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Handheld Specifications Summary
A handheld device is not included in the SCOUT kit. To order an Android device, add it as anaccessory item, or obtain an appropriately rated device directly from the supplier.
For self-supplied handheld devices, we cannot guarantee SCOUT200 compatibility with every
make/model or version of Android. Contact BN TechSupport for a list of tested models, or torequest a test.
Smart Ex 02
DZ2

Smart Ex
02 DZ1Tab ExD2 & DZ2

Tab Ex DZ1

IS930.1

Device Image

Display & Touch

5”, Gorilla glass, glove
touch & wet mode

8” 1920 x 1200, 480 nit, glove
touch & wet mode

8" 1920 x 1200, glove touch

Wireless & GPS

4G/LTE dual SIM,
WiFi 2.4 & 5 GHz, BT 4.2,
NFC, GPS

WiFi 2.4 & 5 GHz, BT 5.0,
NFC, GPS

WiFi 2.4 & 5 GHz, BT 5, NFC,
GPS

Cameras12 MP rear, 5 MP front

13 MP rear, 5 MP front

13 MP rear, 5 MP front

OS

Android 9

Android 11 or later

Android 10

CPU

Qualcomm Snapdragon
450

Exynos 9810, 2.7 GHz,
Octa-core

Qualcomm Snapdragon
SDM660 Octa-Core

Memory3 GB RAM, 32 GB ROM
up to 128 GB microSD

4 GB RAM, 48 GB internal,
up to 1 TB microSD

64 GB internal memory,
up to 128 GB microSD

Battery

Replaceable Li-Ion 4400
mAh

Li-Ion 3.8V, 5050 mAh,
up to 11 hrs

Li-Ion 8400 mAh

Size & Weight

163 x 82 x 22 mm,
380 g

214 x 127 x
10 mm, 610 g

263 x 162 x
26 mm, 1100g

237 x 156 x 22 mm, 890 g

Ruggedness

IP68, MIL-STD-810G,-20 to 60°C

IP68, 1.2m
drop,
-20°C to
+55°C

IP65,
-20°C to+50°C

IP68, -20 to +60°C

Warranty

3 years

3 years

7/16

3 years

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Smart Ex 02
DZ2

Smart Ex
02 DZ1IECEx ATEX

Zone 2/22
Ex ic IIC T4
Gc IP64
Ex ic IIIC
T120°C Dc

DZ2 only:
Zone 1/21
Ex ib IIC T4
Zone 2/22
Gb IP64
Ex ic IIC T5 Gc
Ex ib IIIC
Ex ic IIIC
T120°C Db
T85°C Dc

CSA

Class I,II,III
Div 2,
Groups A ..
G, T4Class I,II,III
Div 1,
Groups A ..
G, T4Complete Specs,
Accessories
SCOUT200
Compatibility

www.ecom-ex.com

Tab ExD2 & DZ2

Tab Ex DZ1

IS930.1

Zone 1/21
Ex ia IIC T6
Gb
Ex ia IIIC
T80°C Db

Zone 1/21,
Ex ib IIC T4 Gb,
Ex ib IIIC T135°C Db

Class I Div 2, Class I,II Div 1,Groups A .. D, Groups A .. G,
T5
T5Class I,II,III Div 1, Groups
A..G, T4

www.ecom-ex.com

www.isafe-mobile.com

These handhelds are WIFI
models, so they do notsupport USB Tethering. This
prevents them from using
USB Direct comms with
System 1.
Available comms methods:
WiFi, File mode, Remotecomms.
To use USB Tethering, order
WWAN version direct from
Pepperl & Fuchs.

Numerous accessories are available for these handhelds, direct from each handheld
manufacturer (e.g. charging stations, belt-clips for the smartphones, and cases for thetablets). You can find the accessories pages on the handheld manufacturers websites.

8/16

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Cyber Security
App Updates

The SCOUT220IS and SCOUT240IS solutions
include an industrial handheld device that runs
the Android operating system and uses USB,
cellular, Bluetooth and Wi-Fi forcommunication. These solutions have featuresthat reduce but do not eliminate cyber security
risks.

The Bently Nevada development teamperiodically updates the S1 Collector App withnew features and enhancements. Built into theapp is the new firmware to also automatically
update the SCOUT200 data collector. These
updates are free to all SCOUT200 owners who
have a current Maintenance and Support
agreement (M&S) for their System 1 software.

We strongly recommend implementing
additional cyber security protections such as:nnnnControlling user access
Scanning for viruses
Applying timely updates
Using mobile device management
software

The updates can be obtained from either ofthese locations:
1. Flexnet Licensing Portal > System 1 >System 1 Collector App.
2. Google Playstore > Search "S1 Collector"

The recommended measures are atyour discretion and responsibility.
Some residual risk may remain.
For details, see the SCOUT200_Series Cyber
Security Hardening Guideline (document
131M1304) at BNTechSupport.com.

9/16

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

RED/EMC

Compliance and Certifications
This section applies only to the Data
Collector, not the handheld device.

EN 61000-6-2
EN 61000-6-4
EN 300 328
EN 301 489-1
EN 301 489-17
EN 61326-1
2014/53/EU
2014/30/EU

This device complies with Industry Canada
license-exempt RSS standard(s) and complies
with part 15 of the FCC Rules, Class A device.
Operation is subject to the following two
conditions:

ATEX/IECEx

1. This device may not cause harmfulinterference.

II 1 (1) GD
II 3 (3) GD
Ex ia [ia] IIC T4 Ga
Ex ic [ic] IIC T4 Gc
Ex ia [ia Ga] IIIB T135°C Da
Ex ic [ic Gc] IIIB T135°C Dc

2. This device must accept any interference
received, including interference that may
cause undesired operation.

CSA/NRTL/C

Changes or modifications not
expressly approved by Bently Nevada
could void the user's authority tooperate the equipment.

Class I Division 1 Groups ABCDClass I Division 2 Groups ABCDClass II Division 1 Groups FG
Class II Division 2 Groups FG
Class I Zone 0, Ex/AEx ia [ia] IIC
T4 Ga
Class I Zone 2, Ex/AEx ic [ic] IIC
T4 Gc
Zone 20, Ex/AEx ia [ia Ga] IIIB
T135°C Da
Zone 22, Ex/AEx ic [ic Gc] IIIB
T135°C Dc

This equipment generates, uses, andcan radiate radio frequency energy
and, if not installed and used inaccordance with the instructions,
may cause harmful interference toradio communications. However,there is no guarantee thatinterference will not occur in aparticular installation.

T4 @ Ta = -20°C to +50°C
The installation must comply
with the SCOUT200 Hazardous
Location Installation Control
Drawing (document 111M0029).

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

ROHS

2011/65/EU

The above ratings include the following areas:n

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

nIntrinsically safe Zone 0, 20 and Division 1
Lower level protection of Zone 2, 22 andDivision 2

Only the following restricted sensor types are
certified for intrinsically safe use:
n

10/16

SCOUT-powered sensors such as ICP/IEPE
accelerometers

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheetn109M0226 Rev. AA

Self-powered tachometers such asbattery-powered handheld laser
tachometers

You can use other types of sensors such as
buffered outputs from instrumentation racks
only in lower rated Zone 2, Division 2 areas.
Refer to SCOUT200 Hazardous Location
Installation Control Drawing (document
111M0029) for details.
For the optional dust certification(Class II Groups F and G, and Zones 20
and 22), special dust-certified
sensors are required. If this isrequired, inform your Bently Nevada
sales professional so that the correct
sensor can be substituted.
The overall hazardous system rating
is defined by the minimum rating of
the data collector, industrial
handheld, and accessories.

11/16

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Part Number

Ordering Information
For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).
For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

SCOUT220IS-AA
A: Data Collector Approval Options
05

Multi Approvals (CSA, IECEx, ATEX)The SCOUT220-IS basic kit does
NOT include a handheld device, nor a
System 1 software license. They areordered separately under Optional
Accessories below.

Part Number

Description

Qty

Description

Qty

AM3100T2Z0 or
200350*Accelerometer, top exit

1

AS3100S2Z0 or
200350*Accelerometer, side exit

1

151M8244-01 Accelerometer cable,
coiled, safety breakaway

2

113M5585

Accelerometer magneticbase

2

109M2385012

BNC to silver-LEMO cable.
Connects rack buffered
outputs to CH1/CH2 inputs.

2

110M8172012

BNC to black-LEMO cable.
Connects rack buffered
Keyphasor to Tach input.

1

109M238401

Anti-static neck strap with
Sensor Keeper

1

117M1446

Belt Clip1

112M4551

Hard carrying case1

109M1269

SCOUT200 Series User
Guide

1
1

S1 Collector app

1

SCOUT220IS Data
Collector

1

109M1270-01 SCOUT200 Series Quick
Start Guide

108M4044

AC power adapter

1

PLUS0230

Category A power plug
(USA / Canada)

1

BentlyManualsDVD

PLSA0241

Category D power plug
(South Africa / India)

1

PLAU0228

Category M power plug
(Australia / New Zealand /China)

1

PLHK0245

Category G power plug
(Hong Kong / UK)1

PLEU0229

Category C Power plug
(Europe)1

Contains the SCOUT220IS
and SCOUT240IS ProductDatasheet (109M0226) ,the SCOUT200 Series
Quick Start Guide(109M1270), and the
SCOUT200 Series User
Guide (document
109M1269).

* Kit items with multiple part numbers listed have
limited regional availability due to certification
requirements.

12/16

1

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Part Number

Description

Qty

AS3100S2Z0 or
200350*Accelerometer, side exit

1

151M8244-01 Accelerometer cable,
coiled, safety breakaway

4

113M5585

Accelerometer magneticbase

4

109M2385012

BNC to silver-LEMO cable.
Connects rack buffered
outputs to CH1/CH2 inputs.

2

110M8172012

BNC to black-LEMO cable.
Connects rack buffered
Keyphasor to Tach input.

1

113M9350

Channel 2-4 breakout
cable

1

Qty

109M238401

Anti-static neck strap with
Sensor Keeper

1

S1 Collector app

1

117M1446

Belt Clip

1

SCOUT240IS
Data Collector

1

112M4551

Hard carrying case1

108M4044

AC power adapter

1

109M1269

SCOUT200 Series User
Guide

1

PLUS0230

Category A power plug
(USA / Canada)1

109M1270-01 SCOUT200 Series Quick
Start Guide

1

PLSA0241

Category D power plug
(South Africa / India)

1

PLAU0228

Category M power plug
(Australia / New Zealand /China)

1

Bently_ManualsDVD

PLHK0245

Category G power plug
(Hong Kong / UK)1

PLEU0229

Category C Power plug
(Europe)1AM3100T2Z0 or
200350*Accelerometer, top exit

3

Metrology certification can be requested whenan order is placed, however this service ischarged. Standard test data can be requested
for free, but is not evidence of calibration.
To request a periodic metrology calibration,
contact Bently Nevada Tech Suppport.

SCOUT240IS-AA
A: Data Collector Approval Options
05

Multi Approvals (CSA, IECEx,
ATEX)
The SCOUT240IS basic kit does
NOT include a handheld device, nor a
System 1 software license. They areordered separately under Optional
Accessories below.

Part Number

Description

Contains the SCOUT220IS
and SCOUT240IS ProductDatasheet (109M0226),
the SCOUT200 Series
Quick Start Guide(109M1270), and the
SCOUT200 Series User
Guide (109M1269)* Kit items with multiple part numbers listed
have limited regional availability due tocertification requirements.

13/16

1

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Other Accessories

Metrology certification can be requested whenan order is placed, however this service ischarged. Standard test data can be requested
for free, but is not evidence of calibration.

109M2389

To request a periodic metrology calibration,
contact Bently Nevada Tech Suppport.

108M4069

Optional Accessories

1.2-meter cable for laser Tach
109M2387
Laser Tach powered by
SCOUT220-IS and SCOUT240-IS,
for Zone 2.
Order cable separately.

Hand Held Devices

CBL50216

5-meter cable for laser Tach
108M4069.

122M9408

Monarch Compact Smart Laser
Sensor (CSLS). Higher powered(3 mW) but larger size (2.6" x2.4" x 2") and no Zone rating.
Order cable separately.

125M209901

Cable for CSLS laser, 1.5 m

146M8212-01

Smart-Ex 02 DZ2(North America &Europe)146M8216-01

Smart-Ex 02 DZ1(North America &Europe)175M2586-01

Tab-Ex D2 North
America

145M3278-01

Tab-Ex DZ2 North
America

113M5529-01

Reflective tape
One roll, 60 cm (23.7 in)

146M8220-01

Tab-Ex DZ2 Europe

VBMR0222

Pair of safety rings for attaching
strap to data collector

151M7906-01

Tab-Ex DZ1 North
America

138M7749

Accelerometer cable, 4 ft,
straight, safety breakaway

151M7907-01

Tab-Ex DZ1 Europe

157M5734-01

iSafe IS930.1 North
America

AP3500T2Z1

Accelerometer, top exit , 500
mV/g, 10 g pk, 0.2 Hz - 2.3 kHz,Div 1 & Zone 1 rated

158M0806-01

iSafe IS930.1 Europe

AP3500S2Z1

Accelerometer, side exit , 500
mV/g, 10 g pk, 0.2 Hz - 3.7 kHz,Div 1 & Zone 1 rated

To obtain a handheld device
suitable for a location not listed
above, contact a handheld device
distributor within the destination
country.

Accessory Kits
Balancing Kit - 161M0958
Part Number

Other Accessories
3071/01

System 1 software and license

109M2387

Handheld, battery-poweredlaser tach for IS zone 0. Order
cable separately.

14/16

Description

Qty

113M5529-01 Reflective tape One roll,
60 cm

1

LASA0315

Laser Tach Kit Zone 2
rated

1

CBL50216

Laser cable Five meters

1

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
DatasheetPart Number

Description

109M0226 Rev. AA

Part Number

Qty

MAGA0063

Laser magnetic stand1

159M9583*

Sensor extension cable,
5m

2

CBBL0026

Carrying case for the kit

1

156M6204

substituted with the CB5G0024 and CB5R0025
cables and two 161M2913 adapters.

Part Number

Zone 2 Laser Tach Kit - LASA0315
Description

108M4064

Laser Tacho Holder

1

108M4066

Circlips - 20Mm Stainless1

108M4067

Arp115 Oring

2

108M4069

Laser Tach Zone 2 rated

1

Triaxial sensor,
100 mV/g ± 10%,
0.6 Hz .. 10 kHz,27 µg/√Hz noise @ 10 Hz

156M6180

Cable for SCOUT200 to triax
155M3055,
1.5 m, straight,
safety breakaway

154M1355

Magnet, 2 pole, 35 mm
diameter, 22 mm high,
23 kg pull, ¼-28 female thread,
triax alignment notch

Language Specific Warning Labels

Triaxial sensor kits are certified forhazardous area use in limited regions(NAM, EU, etc). Other regions mayneed to source locally. Contact
TechSupport for suggested
manufacturer part numbers.

157M0806-01 (Zone 2, Class 1 Div 2)
Description

156M6189

Triaxial sensor,
100 mV/g ± 20%,
0.5 Hz .. 5 kHz,
8 µg/√Hz noise @ 10Hz

157M0852-01

Cable for SCOUT200 to triax
156M6189,
1.5 m, straight,
safety breakaway

Description

155M3055

Qty

Triax Sensor Kits

Part Number

Magnet, 2 pole, 38 mm
diameter,
32 mm high, 25 kg pull,¼-28 female thread

157M0809-01 (Zone 0, CSA, IECEx, ATEX)* Depending on availability, this cable may bePart
NumberDescription

15/16

110M9913

Magnet base label

110M9914

Sensor keeper side label

110M9915

Sensor keeper front label

111M7144

Hard carrying case label

SCOUT220-IS and SCOUT240-IS Portable Data Collectors
Datasheet

109M0226 Rev. AA

Copyright 2023 Baker Hughes Company. All rights reserved.

Bently Nevada, COMMTEST, Keyphasor, M2, SCOUT, System 1 and Orbit Logo are registered trademarks of Bently
Nevada, a Baker Hughes business, in the United States and other countries. The Baker Hughes logo is a trademark of
Baker Hughes Company. All other product and company names are trademarks of their respective holders. Use of
the trademarks does not imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

16/16