SEIFERTERESCO 65 MF4Portable air-cooled 300 kV X-ray generator with 3 mm focal spot
The ERESCO 65 MF4 is a very reliable, portable 300 kV industrial X-ray generator also for use outdoors in the field andunder challenging environmental conditions.
The ERESCO 65 MF4 offers a material penetration of up to65 mm steel with 10 minutes exposure time.*
The X-ray tube head consists of a sturdy metal-ceramic
X-ray tube and high voltage generator integrated into arugged compact housing. The air-cooled 300 KV/ 900 W
end-grounded X-ray tube is thermally protected for extended life.
The high voltage generator generates high-stability constant potential voltage at higher efficiency than conventional line frequency equipment.

Additional Features• Rugged construction with proven reliability• Extended operating range from 5 kV to 300 kV enables
flexible use• Gas insulated high voltage section reduces size & weight• End-grounded, Be-window, metal-ceramic X-ray tube
• DC voltage operation• Modern power electronics provides high stability• On-board real time clock supports fully automatic
warm-up• Large 40° × 60° emergent beam
• Optimized air-flow cooling for efficient heat dissipation• 100 % duty cycle at 30° Celsius (86° F)• Built-in fail-safe warning lamp
• CE certification according to EMC and Low Voltage
Directives• French Standard NFC 74 100 compliant• BfS certification (PTB approval)• Water / dust resistant in accordance with protection
class IP65
• Produced under ISO 9001 certified quality management
systemStandard Delivery Scope
ERESCO generator unit from high voltage generator andX-ray tube or optionally as a system with "ERESCO MF4Control" control unit

The performance of the X-ray tube and the high voltage
generator is continuously monitored for protection andoperational safety. An on-board real time clock facilitates
fully automatic warm-up routines for optimized and timesaving tube conditioning. The flexible connecting cable isheavy-duty and light weight.
The ERESCO series can also be controlled by PC or laptop
and is compatible with the Waygate Technologies software
Rhythm Insight RT.

* When the focus-film distance is 700 mm (27.5 "), 65 mm (2.56") Fe penetration andAGFA STRUCTURIX D7 film with blackening 2.0

Technical Data
X-ray tube typekV range
Target angle

300 kV metal-ceramic with Tungsten (W) anode5 - 300 kV (acc. to rating chart)
20°

Inherent filtration

0.8 mm ± 0.1 mm, Be

Emergent beam

40° × 60°, elliptical

Focal spot size (EN 12 543)

3.0 mm (~1.5 IEC 336)

Tube current

0.5 - 6 mA (3.0 mA /300 kV)

Anode temperature, max.

100° C

Current and voltage stability±1%

Continuous rating

900 W

Duty cycle (δ amb = 30° C)
Power input, max.
Weight (exluding hand rings)Tube head diameter
Protection class

Options• Carrying cradle• Diaphragm / collimator
attachments• Laser centering device• Transport and positioning cart
• Fail-safe flashing warning lamp
• Stands and mounting kits
• Aluminium transport box

100 % (up to 60 min continuous operation)
2.0 kVA
37.8 kg (83.3 lbs)
290 mm (11.42”)
225 mm (7.87”) (without rings)
IP65

Operating Temperature-20° C to +50° C

Storage Temperature-30° C to +80° C

drawings not true to scale

Waygate Technologies
Bogenstr. 41 • 22926 Ahrensburg / Germany
Tel.: +49 4102 807 0 • Fax: +49 4102 807 189 • E-Mail: xray.info@bakerhughes.com
Copyright 2023 Baker Hughes Company. This material contains one or more registeredtrademarks of Baker Hughes Company and its subsidiaries in one or more countries. All
third-party product and company names are trademarks of their respective holders.
BHFF30177EN (05/2023)

waygate-tech.com