DPS5000 CANBus
Digital pressure sensing platform
The new DPS5000 CANBus from Druck integrates digitalelectronics to the high performance UNIK 5000 Pressure
Sensing Platform to provide performance levels unmatchedby traditional analogue sensors. It uses CANopen protocol,
enabling fully compensated readings of pressure andtemperature to be communicated alongside control of many
functions of the device.

Features

High quality• Customer set filter settingWith 40 years of pressure measurement experience, ourfield-proven Druck technology is at the heart of the newplatform, resulting in a range of high quality, high stabilitypressure sensors.

Expertise
We have the people and the knowledge to support yourneeds for accurate and reliable product performance; ourteam of experts can help you make the right sensor selection,guiding you and providing the help and tools you need. It
is important that you ensure that the sensor materials andperformance selected are suitable for your application.

• Ranges from 200 mbar to 700 bar• Total accuracy to ±0.1 % FS
• Stainless steel construction• Wide temperature range -40 °C to 125 °C• Fast update rate 1 ms• Baud rate upto 1000k bit/s• Excellent long-term stability

DPS5000 CANBus specifications
Operating pressure ranges

Output/communications

Gauge ranges

CANopen V2.0B

Any range from 200 mbar to 100 bar (3 to 1500 psi)

CAN setting options (select when ordering)

Ranges up 70 bar are enabled in negative gauge to therange or a maximum of -1 bar (-15 psi)

Node ID: Can be set to any integer between 2 and 127(default setting is 2)

Absolute ranges

Baud rate: Can be set to 10k, 20k, 50k, 125k, 250k, 500k, 800k,1000k (default setting is 250k)

Any range from 700 mbar to 700 bar (10 to 10000 psi)
All ranges are zero based
Wet Dry differential
Any range from 200 mbar to 35 bar (3 to 500 psi)
All units are bidirectional
Maximum line pressure: 70 bar
Wet Wet differential
Any range from 700 mbar to 35 bar (10 to 500 psi)
All units are bidirectional
Maximum line pressure: 70 bar
Ordering
When ordering, specify maximum working pressure.
The lower pressure limit will be specified inline with rules
stated above.
Over pressure
At least 2 × Pressure range with negligible calibration change
For differential versions, the negative side must not exceed its
positive side by more than:
2 × Pressure range up to a maximum of 15 bar
Containment pressure
4 × Pressure range for gauge ranges(200 bar (3000 psi) maximum)
200 bar (3000 psi) for absolute ranges below 100 bar(1500 psi)

Power on time
500 ms to acquisition from power on
Insulation resistance>100 MΩ at 500 Vdc performance

Performance
Pressure performance
Accuracy over the calibrated temperature range including
zero and span setting and the effects of non-linearity,hysteresis and repeatability
Gauge/absolute/wet dry differential
A3 Premium: ±0.1 % FS over -20 to 80 °C (-4 to 176 °F)
A2 Improved: ±0.2 % FS over -40 to 125 °C (-40 to 257 °F)
Wet wet differential
Premium: ±0.2 % FS over -20 to 80 °C (-4 to 176 °F)
Improved: ±0.4 % FS over -40 to 125 °C (-40 to 257 °F)
Increases pro-rata for pressures below 700 mbar
Temperature performance
Accuracy over the calibrated temperature range 		±3 °C (6 °F) with 0.1 °C (0.2 °F) Resolution
Long term stability±0.05 % FS/year typical1200 bar (17000 psi) for absolute ranges above 100 bar(1500 psi)

±0.1 % FS maximum at reference conditions

For differential versions, the negative side must not exceed its
positive side by more than:Line pressure effects (differential sensors only)

2 × Pressure range up to a maximum of 15 bar
SupplyVoltage: 5 Vdc to 32 Vdc
Current: <30 mA

Increases pro-rata below 700 mbar (10 psi)
Zero shift: <±0.03% span/bar
Span shift: <±0.03% span/bar
Effects increase pro-rata for ranges below 700 mbar

Physical specifications
Environmental protection

Wiring details

IP67

Electrical connector

Operating temperature range-40 °C to 125 °C (-40 °F to 257 °F)

M12x1

Pressure media
Fluids compatible with stainless steel 316L and Hastelloy C276

(5-PIN)

Statement in accordance with the European pressure
equipment directive
Enclosure materials
Stainless steel 316L (body)

BAYONET (MIL-C-26482)

Pressure connectors• PA G1/4 female• PB G1/4 male flat

PinFunction

1-2+ VE Supply

3- VE Supply/CAN OV

4CAN Hi

5CAN LoA+ VE SupplyB

CAN HiCCAN LoD- VE Supply

E-F

CAN OV

General certifications• PE 1/4 NPT female

RoHS 2002/95/EC• PF 1/4 NPT male
• PJ M14×1.5 60° Int Cone

CRN Certified OF 13650.513467890YTN for pressure ranges up
to and including 100 bar• PZ M10×1 80° Int ConeCE conformity

Other connectors may be available. Contact Druck to discuss
your requirement.

Pressure Equipment Directive 97/23/EC: Sound Engineering
Practice

Electrical connector

EMC directive 2004/108/EC
BS EN 61326-1:2013

Option code

Description

IP ratingGM12x1 5-PIN-BS EN 61000-6-1: 2007

6BAYONET MIL-C-26482-BS EN 61000-6-3: 2007 + A1:2011

Ordering information1) Select part number

Main product familyDPSDigital Output Transducer
55000 series				Diameter and material		
025mm stainless steel						Electrical connection			GM12x1 5 Pin			6MIL-C-26482								Electronics option				CCANBus										Compensated temperature range					
TB-20 to 80 °C (See note 2)					
TD-40 to 125 °C (See note 3)												Accuracy						A2Improved						A3Premium														Calibration							CCFull thermal																Hazardous area certification								H0None																		Pressure connector									PAG1/4 female (See note 1)									PBG1/4 male Flat									PE1/4 NPT female (See note 1)									PF1/4 NPT male									PJM14x1.5 60° Int cone									PZM10x1 80° Int cone

DPS50GC-TB-A3-

CC-

H0-PA

2) Specify maximum working pressure unit and reference. Available units are:SymbolDescription

SymbolDescription

Symbolmbar

millibar

cmH2O

centimeters water

Kgf/cmpsi

pounds/square inch

mH2O

meters water

atm

hPa

hecto Pascal

ft H2O

feet waterbar

PaKpa

bar

Pascal

Kilo Pascal

mmH2O

in H2O
mmHg

millimeters waterinches waterinHg

2Torrmillimeters mercury3) Specify node id and baud rate (Default: node id-2, baud rate-250k)
Notes:
Note 1: Choose this pressure connector for range over 100 bar (1500 psi).
Note 2: Choose A3-Premium accuracy with this option.
Note 3: Choose A2-Improved accuracy with this option.

Typical order examples
DPS 50GC - TB - A3 - CC - H0 - PZ 700 mbar gauge Node id - 2, Baud rate - 250k
DPS 50GC - TD - A2 - CC - H0 - PA 300 mbar wet dry differential Node id - 2, Baud rate - 500k

Descriptioninches mercury
Kilograms force/
centimeters squareatmospheretorrMechanical drawings

Note: All dimensions in millimeters.

Mechanical drawings

Note: All dimensions in millimeters.

Copyright 2021 Baker Hughes Company. All rights reserved.

920-684B

BHCS38664 (03/2021)

druck.com