SEIFERTERESCO 52 MF4-CL

Portable air-cooled 300 kV panoramic X-ray generator
The ERESCO 52 MF4-CL is a very reliable, portable 300 kV
industrial X-ray generator also for use outdoors in the field
and under challenging environmental conditions.
The ERESCO 52 MF4-CL offers a material penetration of up
to 52 mm steel with 10 minutes exposure time.*
The X-ray tube head consists of a sturdy metal-ceramic
X-ray tube and high voltage generator integrated into arugged compact housing. The air-cooled 300 KV/ 600 W
end-grounded X-ray tube is thermally protected forextended life.
The high voltage generator generates high-stability constant potential voltage at higher efficiency than conventional line frequency equipment.

Additional Features• Rugged construction with proven reliability• Extended operating range from 5 kV to 300 kV enables
flexible use• Gas insulated high voltage section reduces size & weight• End-grounded, metal-ceramic X-ray tube
• DC voltage operation• Modern power electronics provides high stability• On-board real time clock supports fully automaticwarm-up• Large 38° × 360° emergent beam
• Optimized air-flow cooling for efficient heat dissipation• 100 % duty cycle at 30° Celsius (86° F)• Built-in fail-safe warning lamp
• CE certification according to EMC and Low Voltage
Directives• Water / dust resistant in accordance with protection
class IP65
• Produced under ISO 9001 certified quality management
systemStandard Delivery Scope
ERESCO generator unit from high voltage generator andX-ray tube or optionally as a system with "ERESCO MF4Control" control unit

The performance of the X-ray tube and the high voltage
generator is continuously monitored for protection andoperational safety. An on-board real time clock facilitates
fully automatic warm-up routines for optimized and timesaving tube conditioning. The flexible connecting cable isheavy-duty and light weight.
The ERESCO series can also be controlled by PC or laptop
and is compatible with the Waygate Technologies software
Rhythm Insight RT.

* When the focus-film distance is 700 mm (27.5 "), 52 mm (2.01") Fe penetration andAGFA STRUCTURIX D7 film with blackening 2.0

Technical Data
X-ray tube type

300 kV metal-ceramic with Tungsten (W) cone anodekV range

5 - 300 kV (acc. to rating chart)

Target angle

22°

Inherent filtration

0.4 mm Fe/Ni/Co + 3 mm Al

Emergent beam

38° × 360°

Focal spot size (EN 12 543)

0.5 × 5.5 mm

Tube current

0.5 - 6 mA (2.0 mA /300 kV)

Anode temperature, max.

100° C

Current and voltage stability±1%

Continuous rating

600 W

Duty cycle (δ amb = 30° C)

100 %Power input, max.

1.4 kVA

Weight (exluding hand rings)

Options• Carrying cradle• Diaphragm / collimator
attachments• Laser centering device• Transport and positioning cart
• Fail-safe flashing warning lamp
• Stands and mounting kits
• Aluminium transport box• Crawler Integration Kit

33.8 kg (74.5 lbs)

Protection class

IP65

Tube head diameter

225 mm (7.87”) (without rings)

Operating Temperature-20° C to +50° C

Storage Temperature-30° C to +80° C

ERESCO 52 MF4-CL
Exposure Diagram

Belichtungsdiagramm

FFD = 700 mm (27.5") D7Pb D = 2.0

FFA = 700 mm D7Pb S = 2,050VV0k
18V0k0k
16kV

14kV

12080 kV30100200kV4022,5203015mA x min

10 min2,0 mAV0k17125 min
2,0 mA

109873 min
2,0 mA

654,5432005100.25

150.5

200.75

25

1.030

35

1.25

40

1.545

1.75

50

52

2.0mm Feinch Fe

Rating chart

P = 600 W

drawings not true to scale

Waygate Technologies
Bogenstr. 41 • 22926 Ahrensburg / Germany
Tel.: +49 4102 807 0 • Fax: +49 4102 807 189 • E-Mail: xray.info@bakerhughes.com
Copyright 2023 Baker Hughes Company. This material contains one or more registeredtrademarks of Baker Hughes Company and its subsidiaries in one or more countries. All
third-party product and company names are trademarks of their respective holders.
BHFF30176EN (05/2023)

waygate-tech.com