UNIK5600/5700
Marine certified
pressure sensing
platform

The new UNIK 5600/5700 carries marine certification for
most zones on-board ship, as well as Intrinsically Safe
certifications. Marine approval means UNIK 5000 complies
with International standards, regulations and Marine Law.
The use of Druck silicon technology and analogue circuitry
enables best in class performance for stability, low powerand high frequency response. The platform enables
you to build up your own sensor to match your precise
needs. This high performance, configurable solution topressure measurement employs modular design and leanmanufacturing techniques to offer:

High quality

With 40 years of pressure measurement experience, ourfield-proven Druck silicon technology is at the heart of
the new platform, resulting in a range of high quality, highstability pressure sensors.

Bespoke as standard

Custom-built from standard components, manufacturingsensors to your requirement is fast and simple; each UNIK
5000 is a “bespoke” pressure sensing solution, but with theshort lead times and competitive pricing you would expectfrom standard products.

Expertise

We have the people and the knowledge to support yourneeds for accurate and reliable product performance; ourteam of experts can help you make the right sensor selection,
guiding you and providing the help and tools you need.

Features• Ranges from 70 mbar (1 psi) to 700 bar (10,000 psi)
(Depending on material option)• Accuracy to ±0.04% Full Scale (FS) Best Straight Line (BSL)• Stainless Steel 316L and Titanium construction options• Frequency response to 3.5 kHz• High over pressure capability• Intrinsically Safe Hazardous Area certification• 4-20 mA output• Multiple pressure connector options• DIN 43650 or fully submersible electrical connection• Operating temperature ranges from –40 to 80°C (-40 to176°F)

5600/5700 specifications
Measurement
Operating pressure ranges
Gauge ranges
Any zero based range between 70 mbar and 70 bar (1 to 1,000psi) (values in psi are approximate)
Sealed gauge ranges
Any zero based range between 10 and 700 bar (145 to 10,000psi) (Titanium option limited to 70 bar)

Supply voltage
7 to 32 Vdc (7 to 28 Vdc in hazardous area operation)
Output4-20 mA
Power-up time
10 msInsulation• 500 Vdc: 100 MW

Absolute ranges
Any zero based range between 100 mbar and 700 bar (1.5 to10,000 psi) (Titanium option limited to 70 bar)

• 500 Vac: < 5 mA leakage current

Differential ranges (Stainless steel option only)

There are two grades of performance specification: Improvedand Premium

Wet/dry
Uni-directional or bi-directional 70 mbar to 35 bar(1 to 500 psi)
Wet/wetUni-directional or bi-directional 350 mbar to 35 bar(5 to 500 psi)
Line pressure: 70 bar max (1000 psi)
Barometric ranges
Barometric ranges are available with a minimum span of350 mbar (5.1 psi)

Performance specifications

Accuracy
Combined effects of non-linearity, hysteresis andrepeatability:
Improved:±0.1% FS BSL

Premium:±0.04% FS BSL

Note: For the barometric pressure range, accuracy is of span, not full scale.
Note: Accuracy is specified for each direction separately for bi-directional
ranges.

Zero offset and span setting

Non zero based ranges
Non zero based ranges are available. Please contact GE todiscuss your requirements

Demountable electrical connector allows access topotentiometers that give at least ±5% FS adjustment(DIN connector, Demountable Depth Cable only)

Over pressure• 10 × FS for ranges up to 150 mbar (2 psi)

Factory set to:
DIN Connector			Depth Cable			
Demountable Depth Cable• 4 × FS for ranges <350 mbar (5 psi)
• 6 × FS for ranges up to 700 mbar (10 psi)
• 2 × FS for barometric ranges• 4 × FS for all other ranges (up to 200 bar for ranges ≤70 bar
and up to 1200 bar for ranges >70 bar)For differential versions the negative side must not exceed
the positive side by more than:• 6 × FS for ranges up to 150 mbar (2 psi)
• 4 × FS for ranges up to 700 mbar (10 psi)
• 2 × FS for all other ranges up to a maximum of 15 bar(200 psi)
Containment pressure
Ranges up to 150 mbar (2 psi) gauge 10 x FS
Ranges up to 70 bar (1,000 psi) gauge 6 x FS
200 bar (2,900 psi) max)
Ranges up to 70 bar (1,000 psi) absolute 200 bar (2,900 psi)
Ranges above 70 bar (1,000 psi) 1,200 bar (17,400 psi)
Differential (-ve port) must not exceed positive port by morethan 6 × FS (15 bar (200 psi) maximum)±0.2% FS
±1.0% FS
±0.2% FSLong term stability±0.05% FS typical (±0.1% FS maximum) per year increasing
pro-rata for pressure ranges below 350 mbar
Temperature effects-10 to +50 °C (14 to +122 °F): ±0.5% FS
Temperature Error Band (TEB)-10 to +50 °C (14 to +122 °F): ±0.5% FS
Temperature error band (TEB)-20 to +80 °C (-4 to 176 °F): ±1.0% FS TEB-40 to +80 °C (-40 to 176 °F): ±1.5% FS TEB
Temperature effects increase pro-rata for pressure ranges
below 350 mbar (5 psi) and are doubled for barometricranges.
Line pressure effects (Differential version only)
Zero shift: <±0.03% span/bar of line pressure
Span shift: <±0.03% span/bar of line pressure
Effects increase pro-rata for differential pressure ranges
below 700 mbar (10 psi).

Physical specifications
Environmental protection• See electrical connector section

Electrical connector
CodenumberDescription

1. 20 bar (300 psi) maximum for depth cable72.	10 bar (150 psi) maximum for demountable depthcable

Pressure media(Stainless steel 316L option)
Fluids compatible with stainless steel 316L and hastelloy C276.
For the wet/dry differential version, negative pressure port:
fluid compatible with stainless steel 316L, stainless steel 304,
pyrex, silicon and structural adhesive.
(Titanium option)
Fluids compatible with grade 2 and 4 titanium.
Enclosure materials
Stainless steel or titanium (body – material option), glassfilled nylon (DIN connector assembly) with rubber seals(nitrile o-rings and silicone gaskets). PTFE (depth cone, ventfilter), PVDF (cable sheath and depth cone -- depth cableassembly), and TPE-U (cable sheath and depth cone -depth cable assembly).
Pressure connector
Available options are• G1/4 female*• G1/4 male Flat
• G1/2 male via adaptor*• 1/4 NPT male• 1/2 NPT male via adaptor*• M20 X 1.5 male
• Depth cone (G1/4 female open face)
Choose connectors marked * for pressure ranges over 70 bar.
Other pressure connectors may be available. Contact Druck
to discuss your requirement.

IPratingZerospan
adjust°C°F

DIN 43650 Form A
demountable-40 to +80-40 to +176

66YNDepth cable-40 to +80-40 to +176

68

NP

Demountable
kynar cable-40 to +80-40 to +176

68

YU

TPE-U depth cable-40 to +80-40 to +176

68

NV

Demountable
TPE-U cable-40 to +80-40 to +176

68

Y

• Hyperbaric pressure:

Operating temperature range-40 to 80°C (-40 to 176°F)
DNV approval temperature class-25 to 70°C (-13 to 158°F)

Max operating temp range

Wiring details
Codenumber7Connector type

DIN 43650 Form A
demountable

NKynar depth cable

PDemountable kynar
depth cable

UTPE-U depth cable

VDemountable TPE-Udepth cable

Electronics option
1+ve Supply

2-ve Supply

3-ECaseRed+ve supply

White-ve supply

Red+ve supply

White-ve supply

Red+ve supply

White-ve supply

Red+ve supply

White-ve supply

CE conformity

RoHS 2011/65/EU
Pressure equipment directive 2014/68/EU
ATEX 2014/34/EU (optional)
EN 60079-0: 2012+A11: 2013
EN 60079-11: 2007
EN 50303: 2000
EMC directive 2014/30/EU
BS EN 61000-6-1: 2007 		Susceptibility - lightindustrial
BS EN 61000-6-2: 2005 		Susceptibility - heavy
industrial
BS EN 61000-6-3: 2007+A1: 2011
BS EN 61000-6-4: 2007+A1: 2011

Emissions - light industrial	Emissions - heavyindustrial

BS EN 61326-1: 2013		Electrical equipment formeasurement, control andlaboratory use
BS EN 61326-2-3: 2013		Particular requirements forpressure transducers

Hazardous area approvals(optional)

IECEx/ATEX Intrinsically Safe ‘ia’ Group IIC
For full certification details, refer to the type-examination
certificates (or approval listings) and Hazardous Area
Installation Instructions.

Marine approvals

Det Norske Veritas (DNV) approvals: TAA00000JY
Location

ClassTemperature

DHumidity

B

Vibration

BEMCBEnclosure 	(DIN plug)(Depth cable)(Demountable depth cable)

C (IP56)
D (IP68 -200 mH2O)
D (IP68 -100 mH2O)

China Classification Society (CCS) approvals:
NJ16T00162
The pressure transmitter is suitable for use in the following
applications:
1. Trim control
By moving fuel, cargo, and ballast around, the trim of theship is maintained
2. A ship can be considered as an industrial site with engines
and machines which have numerous pressure point
measurement requirements.

Ordering information

See the online configuration tool at www.unik5000.com
1) Select model number
Main product variant
PTX 4-20 mA pressure transmitter
Product series
5UNIK 5000		 Diameter and material		 625 mm stainless steel 316L fluid-isolated (marine approved)		 725 mm titanium fluid-isolated (marine approved)			 Electrical connector			 7DIN 43650 Form A demountable (mating connector supplied)		
N
Depth cable			 PDemountable kynar cable			 UTPE-U depth cable			 VDemountable TPE-U cable				Electronics option				24 to 20 mA 2-wire (PTX)				Compensated temperature range					 TA-10 to +50 °C (14 to +122 °F)					 TB-20 to +80 °C (-4 to +176 °F					 TC-40 to +80 °C (-40 to +176 °F)						 Accuracy						 A2Improved						 A3Premium							 Calibration							 CAZero/span data								 Hazardous area approval								 H0None								 H1IECEx/ATEX intrinsically safe ‘ia’ group IIC								 J1IECEx/ATEX/NEPSI IS LLC Note 1									Pressure Connector									PAG1/4 female Note 1									PBG1/4 male flat									PF1/4 NPT male									PHM20x1.5									PNG1/2 male via adaptor Note 1									PR1/2 NPT male via adaptor Note 1									PWDepth cone (G1/4 female open face)									RJM20x1.5 male (8mm bore)									RKM20x1.5 male (14mm bore)								
PTX5672 - TA - A2 - CA - H0 - PA

Typical model number

Ordering notes:
Note 1: Available only with electrical connector options 7, N and P.
Note 2: Select one of these pressure connectors for pressure ranges over 70 bar.

2) State pressure range and units:e.g. 0 to 10 bar, -5 to + 5 psi
Unit options are:Symbolbar		
mbar		
psi		Pa		hPa		kPa		
MPa		
mmH2O
cmH2O		mH2O		
inH2O		ftH2O		mmHg		
inHg		
kgf/cm2atm		
Torr		

Descriptionbar
millibar
pounds/sq. inch
PascalhectoPascal
kiloPascal
MegaPascal
mm watercm watermetres water
inches waterfeet watermm mercury
inches mercury
kg force/sq. cm
atmospheretorr3) State Pressure reference: e.g. gauge
Reference options are:gauge
absolute
barometricsealed gauge
wet/dry differential
wet/wet differential4) State cable length and units: Integer values only, e.g.,1 mcable, 8 ft. cable. Minimum length 1 m (3 ft.) (only requiredon certain electrical connections). Maximum cable length
100 m (300 ft.).
Typical order examples:

PTX5672-TA-A2-CA-H0-PA, 0 to 3500 psi, absolute
PTX57N2-TA-A2-CA-H0-PA, 0-20 mH2O, gauge, 30 m cableMechanical drawings
7(0.27)72(2.83)

15(0.53)

UNIK 5
PRESSU#### 50##
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. GR

MALE PRESSURE CONNECTION

Ø25 (0.98)

HIGH PRESSURE
CONSTRUCTION

22 A/F(0.86)

69.5(2.73)

15.5(0.61)#### 50##
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:Ø25 (0.98)

Ø25 (0.98)

UNIK 5
PRESSUDRUCK LTD. GR

25 TYP(0.98)

60(2.36)

PVDF CABLE DEPTH CABLE

LOW PRESSURE
CONSTRUCTIONØ25(0.98)

15.5 BOTH ENDS(0.61)

OPTIONAL
PRESSURE ADAPTORUNIK 5PRESSU[2] HIGH PRESSURE IS >70 BAR

Ø25 (0.98)

WET/WET & WET/DRY
DIFFERENTIALCONSTRUCTION
79(3.11)

NOTES:[1] ALL DIMENSIONS ARE IN MILLIMETRES (INCHES)#### 50##
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DEPTH CONE
PRESSURE ADAPTOR

72 (2.83)

DRUCK LTD. GR

16(0.62)50 (1.97)
Ø35 (1.38)

28 (1.10)

46(1.81)MARINE DINCopyright 2021 Baker Hughes Company. All rights reserved.

920-597E

BHCS38616		

(06/2021)

druck.com