Broad-Bandwidth Hybrid OpticalParametric Amplifier
Combination of best collinear
and non-collinear OPA features
Ultrashort pulses in NIR (650 –900 nm and 1200 – 2500 nm)

Single-shot – 2 MHz repetition rate

< 100 fs pulse duration

Adjustable spectral bandwidth

Optional long pulse mode forgap-free tunability

ORPHEUS-F typical tuning curves.
Pump: 40 W, 40 µJ, 1000 kHz

ORPHEUS-F typical spectral bandwidth

5000

51000

1500

0.5

800
SignalIdler

0.1

500.051051

Signal-Long
Idler-Long
SHS-Long
SHI-Long5000.01

Signal-Short
Idler-ShortSHS-ShortSHI-Short
1000

1500

Spectral bandwidth, cm-1100

Pulse energy, µJ

Output power, mW

700
600
500
400
300
200
1000.005

0600

700

800

900 1000

1200

1400

1600 1800 2000

Wavelength, nm

10 ̄ ³2000 2500 3000

Wavelength, nm

ORPHEUS-F pulse duration after compression
100SignalIdler

90

Pulse duration, fs

80
70
60
50
40302010

For custom tuning curves visit
http://toolbox.lightcon.com/tools/tuningcurves/

0600

700

800

900 1000

1200

1400

1600 1800 2000

Wavelength, nm

WWW.LIGHTCON.COM

Specifications
MAIN OUTPUT (650 – 900 nm and 1200 – 2500 nm)
Mode of operationTuning range

Long pulse mode

650 – 900 nm (signal)
1200 – 2500 nm (idler)

650 – 1010 nm (signal)1050 – 2500 nm (idler)

Maximum pump power

80 W

Pump pulse energy

10 – 500 µJ> 7% @ 700 nm

Conversion efficiency 2)
Integrated 2H (515 nm) generation efficiency 3)> 35%

Pulse duration before compression 1)< 290 fs
200 – 750 cm-1 @ 650 – 900 nm

60 – 220 cm-1 @ 650 – 900 nm

Pulse duration after compressor 1)< 55 fs @ 800 – 900 nm< 70 fs @ 650 – 800 nm< 100 fs @ 1200 – 2000 nm

n/aCompressor transmission> 65% @ 650 – 900 nm> 80% @ 1200 – 2000 nm

Spectral bandwidth< 2% @ 800 nm

Long-term power stability, 8h 4)
Pulse-to-pulse energy stability, 1 min< 2% @ 800 nm4)

WAVELENGTH EXTENSION OPTIONS (325 – 15000 nm) 5)
325 – 450 nm (SHS)> 1%

325 – 505 nm (SHS)

n/a
> 1%

n/a

525 – 650 nm (SHI)
600 – 650 nm (SHI)
210 – 252 nm (FHS)
263 – 325 nm (FHI)> 0.5%> 0.5%

n/a

n/a> 0.1%

2500 – 15000 nm

See ORPHEUS-MIR

PUMP LASER REQUIREMENTS
Pump laser

PHAROS or CARBIDECenter wavelength1030 ± 10 nm

Maximum pump power

80 W

Repetition rate

Single-shot – 2 MHz

Pump pulse energy

10 – 500 µJ

Pulse duration 6)

180 – 500 fs

ENVIRONMENTAL & UTILITY REQUIREMENTS
Refer to www.lightcon.com
1)

In short pulse mode, broadband pulses are compressed externally. Typical pulse duration before compression: 120 – 250 fs, after
compression: 25 – 70 fs @ 650 – 900 nm, 40 – 100 fs @ 1200 – 2000 nm.

2)Specified as percentage of pump power, before compressor. Conversion efficiency at peak is equal to 10% for signal and idler combined.

3)

At designated output port; not simultaneous to OPA output.

4)

Expressed as NRMSD (normalized root mean squared deviation).

5)

For > 15 µJ pump pulse energy.

6)

FWHM, assuming Gaussian pulse shape.

Drawings

17DANGER: VISIBLE AND/OR INVISIBLE
LASER RADIATION AVOID EYE OR SKINEXPOSURE TO DIRECT, REFLECTED OR
SCATTERED RADIATION

CLASS 4 LASER PRODUCT

190125

ORPHEUS-F drawings

9013534827

177333701300www.phototechnica.co.jp
REV. 240522 | SALES@LIGHTCON.COM

17125351797012560〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067 FAX:048-871-0068

e-mail:voc@phototechnica.co.jp

Femtosecond Lasers | Wavelength-Tunable Sources | OPCPA Systems | Spectroscopy Systems | Microscopy Sources | Applications

Short pulse mode 1)