GP01024D/06/EN/01.15
71280340

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Proline Prowirl 200
FOUNDATION Fieldbus
Vortex flowmeter

Menu/Variable
P

Proline Device
Access status tooling:
Operation
Setup
Diagnostics
ExpertP

Locking status:

P

Access status tooling:
Enter access code:
SystemP

Sensor
Input
OutputCommunication
Application
Diagnostics

Services

Proline Prowirl 200 FOUNDATION Fieldbus

Table of contents

Table of contents
1

Document information . . . . . . . . . . . . . . 4

1.1
1.2
1.3

Document function . . . . . . . . . . . . . . . . . . . . . 4
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using this document . . . . . . . . . . . . . . . . . . . . 4
1.3.1 Information on the document
structure . . . . . . . . . . . . . . . . . . . . . . . 4
1.3.2 Structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . 6
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . 6
1.4.1 Symbols for certain types of
information . . . . . . . . . . . . . . . . . . . . 6
1.4.2 Symbols in graphics . . . . . . . . . . . . . . . 7

1.4

2

Overview of the Expert operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

3

Description of device parameters . . . 11

3.1

"System" submenu . . . . . . . . . . . . . . . . . . . . . 14
3.1.1 "Display" submenu . . . . . . . . . . . . . . . 14
3.1.2 "Configuration backup display"submenu . . . . . . . . . . . . . . . . . . . . . . 27
3.1.3 "Diagnostic handling" submenu . . . . . . 30
3.1.4 "Administration" submenu . . . . . . . . . 50
3.2
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . . 55
3.2.1 "Measured values" submenu . . . . . . . . 56
3.2.2 "System units" submenu . . . . . . . . . . . 69
3.2.3 "Process parameters" submenu . . . . . . 82
3.2.4 "Measurement mode" submenu . . . . . 85
3.2.5 "External compensation" submenu . . 112
3.2.6 "Sensor adjustment" submenu . . . . . . 117
3.2.7 "Calibration" submenu . . . . . . . . . . . 119
3.3"Output" submenu . . . . . . . . . . . . . . . . . . . . . 120
3.3.1 "Pulse/frequency/switch output"
submenu . . . . . . . . . . . . . . . . . . . . . 120
3.4"Communication" submenu . . . . . . . . . . . . . . 139
3.4.1 "Resource block" submenu . . . . . . . . 140
3.5"Analog inputs" submenu . . . . . . . . . . . . . . . 161
3.5.1 "Analog input 1 to 4" submenu . . . . . 161
3.6"Discrete inputs" submenu . . . . . . . . . . . . . . . 210
3.6.1 "Discrete input 1 to 2" submenu . . . . 211
3.7"Analog outputs" submenu . . . . . . . . . . . . . . 239
3.7.1 "Multiple analog output" submenu . . 239
3.8"Discrete outputs" submenu . . . . . . . . . . . . . . 247
3.8.1 "Multiple discrete output" submenu . . 248
3.9"Application" submenu . . . . . . . . . . . . . . . . . 256
3.9.1 "Totalizer 1 to 3" submenu . . . . . . . . 257
3.10 "Diagnostics" submenu . . . . . . . . . . . . . . . . . 262
3.10.1 "Diagnostic list" submenu . . . . . . . . . 265
3.10.2 "Event logbook" submenu . . . . . . . . . 268
3.10.3 "Device information" submenu . . . . . 271
3.10.4 "Sensor information" submenu . . . . . 275
3.10.5 "Data logging" submenu . . . . . . . . . . 275
3.10.6 "Min/max values" submenu . . . . . . . 281
3.10.7 "Heartbeat" submenu . . . . . . . . . . . . 287
Endress+Hauser

3.10.8 "Simulation" submenu . . . . . . . . . . .

288

4Country-specific factory settings . . 294

4.1

SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . .
4.1.1 System units . . . . . . . . . . . . . . . . . .
4.1.2 Full scale values . . . . . . . . . . . . . . .
4.1.3 Pulse value . . . . . . . . . . . . . . . . . . .
US units . . . . . . . . . . . . . . . . . . . . . . . . . . .
4.2.1 System units . . . . . . . . . . . . . . . . . .
4.2.2 Full scale values . . . . . . . . . . . . . . .
4.2.3 Pulse value . . . . . . . . . . . . . . . . . . .

4.2

294
294
294
295
296
296
296
297

5

Explanation of abbreviated units . . 298

5.1
5.2
5.3
5.4

SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . .
US units . . . . . . . . . . . . . . . . . . . . . . . . . . .
Imperial units . . . . . . . . . . . . . . . . . . . . . . .
Other units . . . . . . . . . . . . . . . . . . . . . . . . .

298
299
300
301

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 302

3

Document information

Proline Prowirl 200 FOUNDATION Fieldbus

1

Document information

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the Expert
operating menu.

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Information on the document structure

The document lists the submenus and their parameters according to the structure from theExpert menu (→  8).

ExpertParameter 1

Parameter n
SystemSensor
Input
OutputCommunication
Application
Diagnostics
A0022576-EN

1

Sample graphic

For information on the arrangement of the parameters according to the structure of
the Operation menu, Setup menu, Diagnostics menu (→  262), along with a brief
description, see the Operating Instructions for the device.

4

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Document information

Language
Operation

Language
Parameter 1
Parameter n
Submenu 1
Submenu n

Setup

Device tag
Wizard 1/Parameter 1

Wizard n/Parameter nAdvanced setup

Parameter 1
Parameter n
Submenu 1
Submenu n

Diagnostics

Parameter 1
Parameter n
Submenu 1
Submenu n
A0022577-EN

2

Sample graphic

For information about the operating philosophy, see the "Operating philosophy"
chapter in the device's Operating Instructions

Endress+Hauser

5Document information

Proline Prowirl 200 FOUNDATION Fieldbus

1.3.2

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter nameWrite-protected parameter = Navigation

Navigation path to the parameter via the local display (direct access code)
Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Options

List of the individual options for the parameter• Option 1• Option 2

User entry

Input range for the parameter

User interface

User interface value/data for parameter

Factory setting

Default setting ex works

Additional information

Additional explanations (e.g. in examples):• On individual options• On display values/data• On the input range• On the factory setting• On the parameter function

1.4

Symbols used

1.4.1

Symbols for certain types of information

SymbolMeaning
Tip
Indicates additional information.
Reference to documentation
Reference to page
Reference to graphic
Operation via local display
Operation via operating tool
Write-protected parameter6

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

1.4.2

Symbols in graphics

SymbolMeaning

SymbolMeaning

1, 2, 3 ...

Item numbers

A, B, C, ...

Views

A-A, B-B, C-C, ...

Endress+Hauser

Document information

Sections

7

Overview of the Expert operating menu

2

Proline Prowirl 200 FOUNDATION Fieldbus

Overview of the Expert operating menu

The following table provides an overview of the menu structure of the expert operating
menu and its parameters. The page reference indicates where the associated description of
the submenu or parameter can be found.

 Expert
Direct access (0106)→  11

Locking status (0004)→  12

Access status display (0091)→  12Enter access code (0092)→  14‣ System→  14‣ Display→  14‣ Configuration backup display→  27‣ Diagnostic handling→  30‣ Administration→  50‣ Sensor→  55‣ Measured values→  56‣ System units→  69‣ Process parameters→  82‣ Measurement mode→  85‣ External compensation→  112‣ Sensor adjustment→  117‣ Calibration→  119‣ Output→  120‣ Pulse/frequency/switch output

8→  120

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Overview of the Expert operating menu

‣ Communication→  139
Device address (11061)→  139‣ Resource block→  140‣ Analog inputs→  161‣ Analog input 1 to 4
‣ Discrete inputs→  210‣ Discrete input 1 to 2
‣ Analog outputs→  211→  239‣ Multiple analog output‣ Discrete outputs→  239→  247‣ Multiple discrete output‣ Application→  248→  256

Reset all totalizers (2806)→  257‣ Totalizer 1 to 3

→  257‣ Diagnostics

Endress+Hauser→  161→  262
Actual diagnostics (0691)→  262

Previous diagnostics (0690)→  263

Operating time from restart (0653)→  264

Operating time (0652)→  264‣ Diagnostic list

→  265‣ Event logbook→  268‣ Device information→  271‣ Sensor information→  275‣ Data logging→  275‣ Min/max values→  2819Overview of the Expert operating menu

10

Proline Prowirl 200 FOUNDATION Fieldbus‣ Heartbeat→  287‣ Simulation→  288

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

3

Description of device parameters

Description of device parameters

In the following section, the parameters are listed according to the menu structure of thelocal display. Specific parameters for the operating tools are included at the appropriate
points in the menu structure.

 Expert
Direct access (0106)→  11

Locking status (0004)→  12

Access status display (0091)→  12Enter access code (0092)→  14‣ System→  14‣ Sensor→  55‣ Output→  120‣ Communication→  139‣ Analog inputs→  161‣ Discrete inputs→  210‣ Analog outputs→  239‣ Discrete outputs→  247‣ Application→  256‣ Diagnostics→  262

Direct access

NavigationDescription

Input of the access code to enable direct access to the desired parameter via the local
display. For this reason, each parameter is assigned a parameter number that appears in
the navigation view on the right in the header of the selected parameter.

User entry

0 to 65 535

Endress+Hauser

Expert → Direct access (0106)

11

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

User entry
The direct access code consists of a 4-digit number and the channel number, which
identifies the channel of a process variable: e.g. 0914-1• The leading zeros in the direct access code do not have to be entered.
Example: Input of "914" instead of "0914"• If no channel number is entered, channel 1 is jumped to automatically.
Example: Enter 0914 → Assign process variable parameter• If a different channel is jumped to: Enter the direct access code with thecorresponding channel number.
Example: Enter 0914-3 → Assign process variable parameter

Locking status
Navigation

 Expert → Locking status (0004)

Description

Use this function to view the active write protection.

User interface• Hardware locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display.
In the operating tool all active types of write protection are selected.
If additional write protection is active, this restricts the current access authorizationeven further. The write protection status can be viewed via the Locking statusparameter (→  12).
"Hardware locked" option (priority 1)
The DIP switch for hardware locking is activated on the main electronics module. This
locks write access to the parameters (e.g. via local display or operating tool).
Information on access authorization is provided in the "User roles and associatedaccess authorization" and "Operating concept" sections of the Operations Instructions
for the device.
"Temporarily locked" option (priority 2)
Write access to the parameters is temporarily locked on account of internal processes
running in the device (e.g. data upload/download, reset etc.). Once the internal processing
has been completed, the parameters can be changed once again.

Access status display
Navigation

 Expert → Access stat.disp (0091)

Prerequisite

A local display is provided.

Description

Use this function to view the access authorization to the parameters via the local display.

12

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User interface• Operator• MaintenanceFactory setting

Operator

Additional information

Description

Description of device parameters

If the -symbol appears in front of a parameter, it cannot be modified via the local display
with the current access authorization.
The access authorization can be modified via the Enter access code parameter(→  14).
For information on the Enter access code parameter (→  14), see the "Disabling
write protection via access code" section of the Operating Instructions for the device
If additional write protection is active, this restricts the current access authorizationeven further. The write protection status can be viewed via the Locking statusparameter (→  12).
User interface
Information on access authorization is provided in the "User roles and associatedaccess authorization" and "Operating concept" sections of the Operations Instructions
for the device.

Access status tooling
Navigation

Expert → Access stat.tool (0005)

Description

Use this function to view the access authorization to the parameters via the operating tool.

User interface• Operator• MaintenanceFactory setting

Maintenance

Additional information

Description
The access authorization can be modified via the Enter access code parameter(→  14).
If additional write protection is active, this restricts the current access authorizationeven further. The write protection status can be viewed via the Locking statusparameter (→  12).
Display
Information on access authorization is provided in the "User roles and associatedaccess authorization" and "Operating concept" sections of the Operations Instructions
for the device.

Endress+Hauser

13Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Enter access code
Navigation

Expert → Ent. access code (0092)

Description

Use this function to enter the user-specific release code to remove parameter write
protection on the local display.

User entry

0 to 9 999

Enter access code
Navigation

Expert → Ent. access code (0003)

Description

Use this function to enter the user-specific release code to remove parameter write
protection in the operating tool.

User entry

0 to 9 999

3.1
Navigation"System" submenu
 Expert → System‣ System

3.1.1
Navigation‣ Display→  14‣ Configuration backup display→  27‣ Diagnostic handling→  30‣ Administration→  50"Display" submenu
 Expert → System → Display‣ Display

14

Language (0104)→  15

Format display (0098)→  16

Value 1 display (0107)→  18Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

0% bargraph value 1 (0123)→  18

100% bargraph value 1 (0125)→  19Decimal places 1 (0095)→  19

Value 2 display (0108)→  20Decimal places 2 (0117)→  20

Value 3 display (0110)→  210% bargraph value 3 (0124)→  21

100% bargraph value 3 (0126)→  22

Decimal places 3 (0118)→  22

Value 4 display (0109)→  22

Decimal places 4 (0119)→  23

Display interval (0096)→  23

Display damping (0094)→  24Header (0097)→  24Header text (0112)→  25

Separator (0101)→  25Contrast display (0105)→  26

Backlight (0111)→  26

Access status display (0091)→  26

Language
Navigation

 Expert → System → Display → Language (0104)

Prerequisite

A local display is provided.

Description

Use this function to select the configured language on the local display.

Endress+Hauser

15Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• English• Deutsch *• Français *• Español *• Italiano *• Nederlands *• Portuguesa *• Polski *• русский язык (Russian) *• Svenska *• Türkçe *• 中文 (Chinese) *• 日本語 (Japanese) *• 한국어 (Korean) *• Bahasa Indonesia *• tiếng Việt (Vietnamese) *• čeština (Czech) *Factory setting

English (alternatively, the ordered language is preset in the device)

Format display
Navigation

 Expert → System → Display → Format display (0098)

Prerequisite

A local display is provided.

Description

Use this function to select how the measured value is shown on the local display.

Selection• 1 value, max. size
• 1 bargraph + 1 value• 2 values• 1 value large + 2 values• 4 values

Factory setting

1 value, max. size

Additional information

Description
The display format (size, bar graph etc.) and number of measured values displayed
simultaneously (1 to 4) can be configured. This setting only applies to normal operation.
• The Value 1 display parameter (→  18)...Value 4 display parameter(→  22) parameters are used to specify which measured values are shown on
the display and in which order.
• If more measured values are specified than the display mode selected permits, thenthe values alternate on the device display. The display time until the next change isconfigured using the Display interval parameter (→  23) parameter.
Possible measured values shown on the local display:*

16

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters"1 value, max. size" option
XXXXXXXXX

900.00l/h
A0016529"1 bargraph + 1 value" option
XXXXXXXXX
900.00 l/h

60.00%
A0016530"2 values" option
XXXXXXXXX

900.00l/h

60.00%
A0016531

3"1 value large + 2 values" option
XXXXXXXXX

900.00l/h
60.00%
3
5.98kWh/Nm
A0016532"4 values" option
XXXXXXXXX
900.00 l/h
60.00 %
3
5.98 kWh/Nm
213.94 l
A0016533

Endress+Hauser

17Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Value 1 display

Navigation

 Expert → System → Display → Value 1 display (0107)

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values to be shown on the local display.

Selection• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *• Reynolds number *• Density *• Totalizer 1• Totalizer 2• Totalizer 3

Factory setting

Volume flow

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the first value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  16) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu(→  69).

0% bargraph value 1Navigation

 Expert → System → Display → 0% bargraph 1 (0123)

Prerequisite

A local display is provided.

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 1.

User entry

Signed floating-point number*

18

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

Country-specific:• 0 m³/h• 0 ft³/h

Additional information

Description

Description of device parameters

The Format display parameter (→  16) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  69).

100% bargraph value 1Navigation

 Expert → System → Display → 100% bargraph 1 (0125)

Prerequisite

A local display is provided.

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 1.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter →  294

Additional information

Description
The Format display parameter (→  16) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  69).

Decimal places 1Navigation

 Expert → System → Display → Decimal places 1 (0095)

Prerequisite

A measured value is specified in the Value 1 display parameter (→  18).

Description

Use this function to select the number of decimal places for measured value 1.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Endress+Hauser

19Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Value 2 display

Navigation

 Expert → System → Display → Value 2 display (0108)

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values to be shown on the local display.

Selection

Picklist, see Value 1 display parameter (→  18)

Factory setting

None

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the second value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  16) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu(→  69).

Decimal places 2Navigation

 Expert → System → Display → Decimal places 2 (0117)

Prerequisite

A measured value is specified in the Value 2 display parameter (→  20).

Description

Use this function to select the number of decimal places for measured value 2.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

20

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Value 3 display

Navigation

 Expert → System → Display → Value 3 display (0110)

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values to be shown on the local display.

Selection

Picklist, see Value 1 display parameter (→  18)

Factory setting

None

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the third value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  16) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu(→  69).

0% bargraph value 3Navigation

 Expert → System → Display → 0% bargraph 3 (0124)

Prerequisite

A selection has been made in the Value 3 display parameter (→  21).

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 3.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 m³/h• 0 ft³/h

Additional information

Description
The Format display parameter (→  16) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  69).

Endress+Hauser

21Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

100% bargraph value 3Navigation

 Expert → System → Display → 100% bargraph 3 (0126)

Prerequisite

A selection was made in the Value 3 display parameter (→  21).

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 3.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
The Format display parameter (→  16) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  69).

Decimal places 3Navigation

 Expert → System → Display → Decimal places 3 (0118)

Prerequisite

A measured value is specified in the Value 3 display parameter (→  21).

Description

Use this function to select the number of decimal places for measured value 3.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Value 4 display

Navigation

 Expert → System → Display → Value 4 display (0109)

Prerequisite

A local display is provided.

22

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to select one of the measured values to be shown on the local display.

Selection

Picklist, see Value 1 display parameter (→  18)

Factory setting

None

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the fourth value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  16) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu(→  69).

Decimal places 4Navigation

 Expert → System → Display → Decimal places 4 (0119)

Prerequisite

A measured value is specified in the Value 4 display parameter (→  22).

Description

Use this function to select the number of decimal places for measured value 4.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Display interval
Navigation

 Expert → System → Display → Display interval (0096)

Prerequisite

A local display is provided.

Description

Use this function to enter the length of time the measured values are displayed if thevalues alternate on the display.

User entry

1 to 10 s

Endress+Hauser

23Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

5sAdditional information

Description
This type of alternating display only occurs automatically if the number of measured
values defined exceeds the number of values the selected display format can display
simultaneously.
• The Value 1 display parameter (→  18)...Value 4 display parameter (→  22)
are used to specify which measured values are shown on the display.
• The display format of the displayed measured values is specified using the Format
display parameter (→  16).

Display damping

Navigation

 Expert → System → Display → Display damping (0094)

Prerequisite

A local display is provided.

Description

Use this function to enter the reaction time of the local display to fluctuations in themeasured value caused by process conditions.

User entry

0.0 to 999.9 s

Factory setting

5.0 s

Additional information

User entry
A time constant is entered:• If a low time constant is entered, the display reacts particularly quickly to fluctuating
measured variables.
• On the other hand, the display reacts more slowly if a high time constant is entered.

Header

Navigation

 Expert → System → Display → Header (0097)

Prerequisite

A local display is provided.

Description

Use this function to select the contents of the header of the local display.

Selection• Device tag• Free text

Factory setting

Device tag

Additional information

Description
The header text only appears during normal operation.

24

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

1

XXXXXXXXX

A0013375

1

Position of the header text on the display

Selection• Device tag
Is defined in the Device tag parameter (→  271).
• Free text
Is defined in the Header text parameter (→  25).

Header text

Navigation

 Expert → System → Display → Header text (0112)

Prerequisite

The Free text option is selected in the Header parameter (→  24).

Description

Use this function to enter a customer-specific text for the header of the local display.

User entry

Max. 12 characters such as letters, numbers or special characters (e.g. @, %, /)Factory setting------------

Additional information

Description
The header text only appears during normal operation.
1

XXXXXXXXX

A0013375

1

Position of the header text on the display

User entry
The number of characters displayed depends on the characters used.

Separator



Navigation

 Expert → System → Display → Separator (0101)

Prerequisite

A local display is provided.

Endress+Hauser

25Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Description

Use this function to select the decimal separator.

Selection• . (point)• , (comma)

Factory setting. (point)

Contrast display
Navigation

 Expert → System → Display → Contrast display (0105)

Prerequisite

A local display is provided.

Description

Use this function to enter a value to adapt the display contrast to the ambient conditions(e.g. the lighting or viewing angle).

User entry

20 to 80 %Factory setting

Depends on the display

Additional information

Set the contrast via the push-buttons:• Brighter: Press and hold down the   keys simultaneously.
• Darker: Press and hold down the   keys simultaneously.

Backlight
Navigation

 Expert → System → Display → Backlight (0111)

Prerequisite

Order code for "Display; operation", option E "SD03 4-line, illum.; touch control + data
backup function"

Description

Option for switching the backlight of the local display on and off.

Selection• Disable• Enable

Factory setting

DisableAccess status display
Navigation

 Expert → System → Display → Access stat.disp (0091)

Prerequisite

A local display is provided.

Description

Use this function to view the access authorization to the parameters via the local display.

26

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User interface• Operator• MaintenanceFactory setting

Operator

Additional information

Description

Description of device parameters

If the -symbol appears in front of a parameter, it cannot be modified via the local display
with the current access authorization.
The access authorization can be modified via the Enter access code parameter(→  14).
For information on the Enter access code parameter (→  14), see the "Disabling
write protection via access code" section of the Operating Instructions for the device
If additional write protection is active, this restricts the current access authorizationeven further. The write protection status can be viewed via the Locking statusparameter (→  12).
User interface
Information on access authorization is provided in the "User roles and associatedaccess authorization" and "Operating concept" sections of the Operations Instructions
for the device.

3.1.2"Configuration backup display" submenu

Navigation

 Expert → System → Conf.backup disp

‣ Configuration backup display
Operating time (0652)→  27

Last backup (0102)→  28Configuration management (0100)→  28Comparison result (0103)→  29Operating time
Navigation

 Expert → System → Conf.backup disp → Operating time (0652)

Description

Use this function to display the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The maximum number of days is 9999, which is equivalent to 27 years.

Endress+Hauser

27Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Last backup
Navigation

 Expert → System → Conf.backup disp → Last backup (0102)

Prerequisite

A local display is provided.

Description

Use this function to display the time since a backup copy of the data was last saved to thedisplay module.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Configuration management

Navigation

 Expert → System → Conf.backup disp → Config. managem. (0100)

Prerequisite

A local display is provided.

Description

Use this function to select an action to save the data to the display module.

Selection• Cancel• Execute backup• Restore• Duplicate• Compare• Clear backup data
• Display incompatible

Factory setting

CancelAdditional information

Description
Configuration via the local display is disabled while the action is performed.
For information on the status message in the operating tool, see: Backup stateparameter (→  29)
Selection• CancelNo action is executed and the user exits the parameter.
• Execute backup– A backup copy of the current device configuration in the HistoROM is saved to thedisplay module of the device. The backup copy includes the transmitter data of thedevice.
– The following message appears on local display: Backup active, please wait!
• Restore– The last backup copy of the device configuration is copied from the display module to
the HistoROM of the device. The backup copy comprises the transmitter data of thedevice.
– The following message appears on local display: Restore active! Do not interrupt power
supply!

28

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Duplicate– The transmitter configuration from another device is duplicated to the device using thedisplay module.
– The following message appears on local display: Copy active! Do not interrupt power
supply!
• Compare– The device configuration saved in the display module is compared to the current device
configuration of the HistoROM.
– The following message appears on local display: Comparing files– The result can be viewed in Comparison result parameter (→  29).
• Clear backup data
– The backup copy of the device configuration is deleted from the display module of thedevice.
– The following message appears on local display: Deleting file
• Display incompatible– This option is displayed if the display module is incompatible. All of the other options
are not available. Selection is therefore not possible.
– This option is displayed if it is not possible to save the device and fieldbus data.
– The display module should be updated to the latest software version so that the data
can be saved.
For information on updating the display module to the latest software version, pleasecontact your Endress+Hauser service organization.
HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

Backup state
Navigation

Expert → System → Conf.backup disp → Backup state (0121)

Prerequisite

A local display is provided.

Description

Use this function to view the status of the data backup process.

User interface• None• Store in progress• Restore in progress• Import in progress• Delete in progress• Compare in progress

Factory setting

None

Comparison result
Navigation

 Expert → System → Conf.backup disp → Compar. result (0103)

Prerequisite

A local display is provided.

Endress+Hauser

29Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Description

Use this function to view the last result of comparing the current device configuration to
the backup copy in the display module.

User interface• Settings identical• Settings not identical• No backup available• Backup settings corrupt• Check not done
• Dataset incompatible

Factory setting

Check not done

Additional information

Description
The comparison is started via the Compare option in the Configuration
management parameter (→  28).
Selection• Settings identical– The current device configuration of the HistoROM is identical to the backup copy in thedisplay module.
– If the transmitter configuration of another device has been copied to the device via thedisplay module and the Duplicate option in the Configuration management
parameter (→  28), the current device configuration of the HistoROM only partly
matches the backup copy in the display module: The settings for the transmitter are
not identical.
• Settings not identical
The current device configuration of the HistoROM is not identical to the backup copy inthe display module.
• No backup available
There is no backup copy of the device configuration of the HistoROM in the display
module.
• Backup settings corrupt
The current device configuration of the HistoROM is corrupt or not compatible with the
backup copy in the display module.
• Check not done
The device configuration of the HistoROM has not yet been compared to the backup copyin the display module.
• Dataset incompatible
The backup copy in the display module is not compatible with the device.
HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

3.1.3
Navigation"Diagnostic handling" submenu
 Expert → System → Diagn. handling‣ Diagnostic handling
Alarm delay (0651)30→  31

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters‣ Diagnostic behavior→  32‣ Diagnostic limits→  48

Alarm delay

Navigation

 Expert → System → Diagn. handling → Alarm delay (0651)

Description

Use this function to enter the time interval until the device generates a diagnostic
message.
The diagnostic message is reset without a time delay.

User entry

0 to 60 s

Factory setting

0s

Additional information

Description
This setting affects the following diagnostic messages:• 046 Sensor limit exceeded• 828 Ambient temperature too low• 829 Ambient temperature too high• 832 Electronic temperature too high• 833 Electronic temperature too low• 834 Process temperature too high• 835 Process temperature too low• 841 Flow velocity too high• 841 Sensor range• 844 Sensor range exceeded• 870 Measuring inaccuracy increased• 871 Near steam saturation limit• 872 Wet steam detected• 873 No steam detected• 874 Wet steam detection uncertain• 945 Sensor range exceeded• 946 Vibration detected• 947 Vibration exceeded

Endress+Hauser

31Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus"Diagnostic behavior" submenu
For a list of all the diagnostic events, see the Operating Instructions for the device.
Modifying the diagnostic behavior of a diagnostic event. Each diagnostic event is
assigned a certain diagnostic behavior at the factory. The user can change thisassignment for certain diagnostics events.
The following options are available in the Assign behavior of diagnostic no. xxx
parameters:• Off option
The diagnostic event is ignored; it is neither entered into the Event logbook, nor is adiagnostic message generated.
• Alarm option
The device continues to measure. The signal outputs assume the defined alarm
condition. A diagnostic message is generated.
• Warning option
The device continues to measure. A diagnostic message is generated.
• Logbook entry only option
The device continues to measure. The diagnostic message is entered in the Eventlogbook submenu (→  268) (Event list submenu (→  270)) only and is not
displayed in alternation with the measured value display.
Assignment of the status of a diagnostic event.
The following options are available in the Assign status of diagnostic no. xxx
parameters:• Bad
The status of the input value (PV) is Bad.
• Uncertain
The status of the input value (PV) is Uncertain.
• GoodThe status of the input value (PV) is Good.
Navigation

 Expert → System → Diagn. handling → Diagn. behavior‣ Diagnostic behavior

32

Assign behavior of diagnostic no. 022(0751)→  34

Assign behavior of diagnostic no. 122(0752)→  35Assign behavior of diagnostic no. 350(0756)→  35Assign behavior of diagnostic no. 371(0757)→  36

Assign behavior of diagnostic no. 442(0658)→  36

Assign behavior of diagnostic no. 443(0659)→  36

Assign behavior of diagnostic no. 828(0755)→  37

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Endress+Hauser

Description of device parameters

Assign behavior of diagnostic no. 829(0754)→  37

Assign behavior of diagnostic no. 832(0675)→  37

Assign behavior of diagnostic no. 833(0676)→  38Assign behavior of diagnostic no. 834(0677)→  38Assign behavior of diagnostic no. 835(0678)→  39Assign behavior of diagnostic no. 841(0729)→  39Assign behavior of diagnostic no. 844(0747)→  39Assign behavior of diagnostic no. 870(0726)→  40Assign behavior of diagnostic no. 871(0748)→  40Assign behavior of diagnostic no. 872(0746)→  40Assign behavior of diagnostic no. 873(0749)→  41

Assign behavior of diagnostic no. 945(0750)→  41

Assign behavior of diagnostic no. 947(0753)→  42

Assign status of diagnostic number 022(11041)→  42

Assign status of diagnostic number 122(11042)→  42

Assign status of diagnostic number 350(11000)→  43

Assign status of diagnostic number 371(11014)→  43

Assign status of diagnostic number 828(11015)→  4333

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign status of diagnostic number 829(11001)→  44

Assign status of diagnostic number 832(11002)→  44

Assign status of diagnostic number 833(11003)→  44

Assign status of diagnostic number 834(11004)→  45Assign status of diagnostic number 835(11005)→  45Assign status of diagnostic number 841(11006)→  45Assign status of diagnostic number 844(11007)→  46

Assign status of diagnostic number 870(11008)→  46

Assign status of diagnostic number 871(11009)→  46

Assign status of diagnostic number 872(11010)→  47

Assign status of diagnostic number 873(11011)→  47

Assign status of diagnostic number 945(11012)→  47

Assign status of diagnostic number 947(11013)→  48Assign behavior of diagnostic no. 022 (Temperature sensor defective)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 022 (0751)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to change the diagnostic behavior of the diagnostic message
022 Temperature sensor defective.

34

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Alarm

Additional information

Description of device parameters

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 122 (Temperature sensor defective)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 122 (0752)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to change the diagnostic behavior of the diagnostic message
122 Temperature sensor defective.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 350 (Pre-amplifier defective)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 350 (0756)

Description

Use this function to change the diagnostic behavior of the diagnostic message 350 Preamplifier defective.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Alarm

Additional information

Endress+Hauser

For a detailed description of the options available for selection: →  32

35

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign behavior of diagnostic no. 371 (Temperature sensor defective)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 371 (0757)

Description

Use this function to change the diagnostic behavior of the diagnostic message
371 Temperature sensor defective.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 442 (Frequency output)



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 442 (0658)

Prerequisite

The measuring device has a pulse/frequency/switch output.

Description

Use this function to change the diagnostic behavior of the diagnostic message
442 Frequency output.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 443 (Pulse output)



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 443 (0659)

Prerequisite

The measuring device has a pulse/frequency/switch output.

Description

Use this function to change the diagnostic behavior of the diagnostic message 443 Pulse
output.

Selection• Off• Alarm• Warning• Logbook entry only

36

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting
Additional information

Description of device parameters

Warning
For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 828 (Ambient temperature too low)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 828 (0755)

Description

Use this function to change the diagnostic behavior of the diagnostic message
828 Ambient temperature too low.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 829 (Ambient temperature too high)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 829 (0754)

Description

Use this function to change the diagnostic behavior of the diagnostic message
829 Ambient temperature too high.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 832 (Electronic temperature too high)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 832 (0675)

Description

Use this function to change the diagnostic behavior of the diagnostic message
832 Electronic temperature too high.

Endress+Hauser

37Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 833 (Electronic temperature too low)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 833 (0676)

Description

Use this function to change the diagnostic behavior of the diagnostic message
833 Electronic temperature too low.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 834 (Process temperature too high)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 834 (0677)

Description

Use this function to change the diagnostic behavior of the diagnostic message 834 Processtemperature too high.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

38

For a detailed description of the options available for selection: →  32

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Assign behavior of diagnostic no. 835 (Process temperature too low)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 835 (0678)

Description

Use this function to change the diagnostic behavior of the diagnostic message 835 Processtemperature too low.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 841 (Flow velocity too high)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 841 (0729)

Description

Use this function to change the diagnostic behavior of the diagnostic message 841 Flowvelocity too high.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 844 (Sensor range exceeded)



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 844 (0747)

Description

Use this function to change the diagnostic behavior of the diagnostic message 844 Sensor
range exceeded.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information
Endress+Hauser

For a detailed description of the options available for selection: →  32
39

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign behavior of diagnostic no. 870 (Measuring inaccuracy increased)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 870 (0726)

Description

Use this function to change the diagnostic behavior of the diagnostic message
870 Measuring inaccuracy increased.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 871 (Near steam saturation limit)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 871 (0748)

Prerequisite

In the Select medium parameter (→  85), the Steam option is selected.

Description

Use this function to change the diagnostic behavior of the diagnostic message 871 Near
steam saturation limit.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 872 (Wet steam detected)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 872 (0746)

Prerequisite

The Wet steam detection application package has been enabled.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

40

Use this function to change the diagnostic behavior of the diagnostic message 872 Wet
steam detected.

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

Description of device parameters

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 873 (Water detected)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 873 (0749)

Prerequisite

In the Select medium parameter (→  85), the Steam option is selected.

Description

Use this function to change the diagnostic behavior of the diagnostic message 873 Water
detected.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Off

Additional information

For a detailed description of the options available for selection: →  32

Assign behavior of diagnostic no. 945 (Sensor range exceeded)



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 945 (0750)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to change the diagnostic behavior of the diagnostic message 945 Sensor
range exceeded.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

Endress+Hauser

For a detailed description of the options available for selection: →  32

41

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign behavior of diagnostic no. 947 (Vibration exceeded)



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 947 (0753)

Description

Use this function to change the diagnostic behavior of the diagnostic message
947 Vibration exceeded.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Alarm

Additional information

For a detailed description of the options available for selection: →  32

Assign status of diagnostic number 022 (Temperature sensor defective)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 022 (11041)

Description

Option for changing the status of the diagnostic message 022 Temperature sensor
defective.

Selection• Bad• Uncertain• GoodFactory setting

Bad

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 122 (Temperature sensor defective)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 122 (11042)

Description

Option for changing the status of the diagnostic message 122 Temperature sensor
defective.

Selection• Bad• Uncertain• GoodFactory setting

Good

Additional information

42

Detailed description of the options available for selection: →  32

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Assign status of diagnostic number 350 (Pre-amplifier defective)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 350 (11000)

Description

Option for changing the status of the diagnostic message 350 Pre-amplifier defective.

Selection• Bad• Uncertain• GoodFactory setting

Bad

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 371 (Temperature sensor defective)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 371 (11014)

Description

Option for changing the status of the diagnostic message 371 Temperature sensor
defective.

Selection• Bad• Uncertain• GoodFactory setting

Bad

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 828 (Ambient temperature too low)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 828 (11015)

Description

Option for changing the status of the diagnostic message 828 Ambient temperature too
low.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  32

43

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign status of diagnostic number 829 (Ambient temperature too high)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 829 (11001)

Description

Option for changing the status of the diagnostic message 829 Ambient temperature toohigh.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 832 (Electronic temperature too high)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 832 (11002)

Description

Option for changing the status of the diagnostic message 832 Electronic temperature toohigh.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 833 (Electronic temperature too low)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 833 (11003)

Description

Option for changing the status of the diagnostic message 833 Electronic temperature too
low.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

44

Detailed description of the options available for selection: →  32

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Assign status of diagnostic number 834 (Process temperature too high)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 834 (11004)

Description

Option for changing the status of the diagnostic message 834 Process temperature toohigh.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 835 (Process temperature too low)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 835 (11005)

Description

Option for changing the status of the diagnostic message 835 Process temperature too
low.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 841 (Flow velocity too high)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 841 (11006)

Description

Option for changing the status of the diagnostic message 841 Flow velocity too high.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  32

45

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign status of diagnostic number 844 (Sensor range exceeded)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 844 (11007)

Description

Option for changing the status of the diagnostic message 844 Sensor range exceeded.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 870 (Measuring inaccuracy increased)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 870 (11008)

Description

Option for changing the status of the diagnostic message 870 Measuring inaccuracy
increased.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 871 (Near steam saturation limit)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 871 (11009)

Description

Option for changing the status of the diagnostic message 871 Near steam saturation
limit.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information46

Detailed description of the options available for selection: →  32

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Assign status of diagnostic number 872 (Wet steam detected)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 872 (11010)

Description

Option for changing the status of the diagnostic message 872 Wet steam detected.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 873 (No steam detected)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 873 (11011)

Description

Option for changing the status of the diagnostic message 873 No steam detected.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Assign status of diagnostic number 945 (Sensor range exceeded)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 945 (11012)

Description

Option for changing the status of the diagnostic message 945 Sensor range exceeded.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  32

47

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign status of diagnostic number 947 (Vibration exceeded)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 947 (11013)

Description

Option for changing the status of the diagnostic message 947 Vibration exceeded.

Selection• Bad• Uncertain• GoodFactory setting

Bad

Additional information

Detailed description of the options available for selection: →  32

"Diagnostic limits" submenu
Navigation

 Expert → System → Diagn. handling → Diagn. limits‣ Diagnostic limits
Reynolds number limit (7646)→  48

Steam quality limit (7717)→  49Reynolds number limitNavigation

 Expert → System → Diagn. handling → Diagn. limits → Re number limit (7646)

Prerequisite

For the following order code:"Sensor version", option "Mass flow (integrated temperature measurement)"

Description

Use this function to enter the lower limit value for the Reynolds number. If the Reynolds
number falls short of this limit value, the diagnostic message 870 Measuring inaccuracy
increased is triggered.

User entry

4 000 to 100 000

Factory setting

5 000

Additional information

Limit value
If the Reynolds number falls short of the limit value configured here, the diagnostic
behavior selected in the Assign behavior of diagnostic no. 870 parameter (→  40)
is triggered.

48

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Steam quality limitNavigation

 Expert → System → Diagn. handling → Diagn. limits → SteamQualLimit (7717)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Steam option is selected.
• In the Steam quality parameter (→  116), the Calculated value option is selected.

Description

Use this function to enter the threshold value for the steam quality which, if undershot,
triggers the diagnostic message S872 Wet steam detected.

User entry

80 to 100 %Factory setting

80 %Additional information

Limit value
This limit value has a hysteresis of 5 %, i.e. the diagnostic message is reset at a thresholdvalue of +5 % or if 100 % is reached (for factory setting of 80 % at 85 %).
If the steam quality has dropped below the limit value configured here, the diagnostic
behavior selected in the Assign behavior of diagnostic no. 872 parameter (0746)(→  40) is triggered.

Degrees of superheat limitNavigation

 Expert → System → Diagn. handling → Diagn. limits → Degr.superh.lim. (7737)

Prerequisite

In the Select medium parameter (→  85), the Steam option is selected.

Description

Use this function to enter the threshold value for the degree of superheat which, if
exceeded, triggers the diagnostic message 972 Degrees of superheat limit excceeded.

User entry

0 to 500 K

Factory setting

5K

Additional information

Limit value
This limit value has a hysteresis of 1 K, i.e. the diagnostic message is triggered if thethreshold value +1 K is reached and is reset again when the value drops below thethreshold value.
If the degree of superheat has exceeded the limit value configured here, the diagnostic
behavior selected in the Assign behavior of diagnostic no. 972 parameter istriggered.

Endress+Hauser

49

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

3.1.4"Administration" submenu

Navigation

 Expert → System → Administration‣ Administration‣ Define access code→  50

Restart (10800)→  52

Activate SW option (0029)→  53

Software option overview (0015)→  54

Activate sensor emergency mode
(7712)→  54"Define access code" wizard
The Define access code wizard is only available if operating using the local display. If
you are operating using the operating tool, the Define access code parameter(→  51) is directly in the Administration submenu. The Confirm access code
parameter is not available if you are operating using the operating tool.
Navigation

Expert → System → Administration → Def. access code

‣ Define access code
Define access code→  50Confirm access code→  51

Define access code

NavigationDescription

Use this function to enter a user-specific release code to restrict write-access to theparameters. This protects the configuration of the device against any inadvertent changes
via the local display.

User entry

0 to 9 999

Factory setting

050

Expert → System → Administration → Def. access code → Def. access code

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
The write protection affects all parameters in the document marked with the  symbol.
On the local display, the  symbol in front of a parameter indicates that the parameter iswrite-protected.
Once the access code has been defined, write-protected parameters can only be
modified if the access code is entered in the Enter access code parameter (→  14).
If you lose the access code, please contact your Endress+Hauser Sales Center.
User entry
A message is displayed if the access code is not in the input range.
Factory setting
If the factory setting is not changed or 0 is defined as the access code, the parameters are
not write-protected and the device configuration data can be modified. The user is logged
on in the "Maintenance" role.

Confirm access code

Navigation

Expert → System → Administration → Def. access code → Confirm code

Description

Enter the defined release code a second time to confirm the release code.

User entry

0 to 9 999

Factory setting

0

Further parameters in the "Administration" submenu

Define access code

NavigationDescription

Use this function to enter a user-specific release code to restrict write-access to theparameters. This protects the configuration of the device against any inadvertent changes
via the operating tool.

User entry

0 to 9 999

Factory setting0

Endress+Hauser

Expert → System → Administration → Def. access code (0093)

51

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
The write protection affects all parameters in the document marked with the  symbol.
Once the access code has been defined, write-protected parameters can only be
modified if the access code is entered in the Enter access code parameter (→  14).
If you lose the access code, please contact your Endress+Hauser Sales Center.
User entry
A message is displayed if the access code is not in the input range.
Factory setting
If the factory setting is not changed or 0 is defined as the access code, the parameters are
not write-protected and the device configuration data can be modified. The user is logged
on in the "Maintenance" role.

RestartNavigation

 Expert → System → Administration → Restart (10800)

Description

Use this function to select a manual restart or a manual device reset.

Selection• Uninitialized• Run• Resource• Defaults• Processor• To factory defaults• To delivery settings• ENP restart• To transducer defaults• Factory Default Blocks

Factory setting

Uninitialized

Additional information"Uninitialized" option
The selection does not affect configuration.
"Run" option
The selection does not affect configuration.
"Resource" option
The selection does not affect configuration.
"Defaults" option
All fieldbus function blocks are reset to their factory setting (e.g. Analog input channel to
the Uninitialized option).
"Processor" option
The device is restarted.

52

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters"To factory defaults" option
Advanced fieldbus parameters (fieldbus function blocks, schedule information) and device
parameters are reset to their factory setting.
"To delivery settings" option
Advanced fieldbus parameters (fieldbus function blocks, schedule information) and device
parameters for which a customer-specific default setting was ordered are reset to thiscustomer-specific value.
"ENP restart" option
The parameters of the electronic name plate (ENP = Electronic Name Plate) to be enteredare adopted. The device is restarted.
"To transducer defaults" option
Certain (measured-value specific) device parameters are reset. The parameters of the
fieldbus function blocks remain unchanged.
"Factory Default Blocks" option
Advanced fieldbus parameters (fieldbus function blocks, schedule information) are reset totheir factory setting.

Activate SW option



Navigation

 Expert → System → Administration → Activate SW opt. (0029)

Description

Use this function to enter an activation code to enable an additional, ordered software
option.

User entry

Max. 10-digit string consisting of numbers.

Factory setting0

Additional information

User entry
Endress+Hauser provides the corresponding activation code for the software option
with the order.
NOTICE! This activation code varies depending on the measuring device and the software
option. If an incorrect or invalid code is entered, this can result in the loss of software
options that are already been activated. After commissioning the measuring device: in thisparameter only enter activation codes which Endress+Hauser has provided (e.g. when a
new software option was ordered). If an incorrect or invalid activation code is entered,enter the activation code from the parameter protocol again and contact yourEndress+Hauser sales organization, quoting the serial number of your device.
Example for a software option
Order code for "Application package", option EA "Extended HistoROM"

Endress+Hauser

53Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Software option overview
Navigation

 Expert → System → Administration → SW option overv. (0015)

Description

Displays all the software options that are enabled in the device.

User interface• Extended HistoROM• Mass flow
• Natural gas• Air + industrial gas• Wet steam detection• Heartbeat Verification

Additional information

Description
Displays all the options that are available if ordered by the customer.
"Extended HistoROM" option
Order code for "Application package", option EA "Extended HistoROM""Mass flow" option
Order code for "Sensor version"• For Prowirl D, F, R: option 3 "Mass flow (integrated temperature measurement)"• For Prowirl C, O: option 6 "Mass flow Alloy 718"
"Natural gas" option
Order code for "Application package", option EN "Natural gas""Air + industrial gas" option
Order code for "Application package", option ET "Air+industrial gases (single gas+gas
mixtures)""Wet steam detection" option
Only available for Prowirl F.
Order code for "Application package", option ES "Wet steam detection""Heartbeat Verification" option
Order code for "Application package", option EB "Heartbeat Verification"

Activate sensor emergency modeNavigation

 Expert → System → Administration → Sens. emerg.mode (7712)

Prerequisite

The device has identified an error during verification of the characteristics in the sensor
data storage or electronics module. A diagnostic message of status type F is output.

Description

Use this function to switch on the emergency mode of the sensor to use the backup of thesensor characteristics or main electronics characteristics stored in the HistoROM.

54

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Cancel• Ok

Factory setting

CancelAdditional information

Description

Description of device parameters

This parameter becomes visible if the data in the S-DAT or on-board memory cannot
be read on account of a defect or error. There is a copy of the data on the HistoROM(FT10). If the emergency mode is activated, this copy is used and the device measurecorrectly again at least up until the next device switch-off/switch-on. After switchon/switch-off, the emergency mode would have to be reactivated again. This ensuresthat the client can operate the device until a new spare part arrives.
The status signal of the output diagnostic message changes from F (failure) to M
(maintenance required), the diagnostic behavior changes from Alarm to Warning: M.
The diagnostic message is output until the characteristics in the sensor data storage areagain correct.
Information on what is causing the diagnostic message, and remedy measures, can beviewed by pressing the -button.
Information on status signals and diagnostic behavior: Operating Instructions about
the device, "Diagnostic message" chapter

3.2
Navigation"Sensor" submenu
 Expert → Sensor‣ Sensor

Endress+Hauser‣ Measured values→  56‣ System units→  69‣ Process parameters→  82‣ Measurement mode→  85‣ External compensation→  112‣ Sensor adjustment→  117‣ Calibration→  11955

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

3.2.1"Measured values" submenu

Navigation

 Expert → Sensor → Measured val.

‣ Measured values‣ Process variables→  56‣ Totalizer→  66‣ Output values→  67"Process variables" submenu
Navigation

 Expert → Sensor → Measured val. → Process variab.

‣ Process variables

56

Volume flow (1838)→  57

Corrected volume flow (1850)→  57Mass flow (1847)→  58Flow velocity (1865)→  59Temperature (1851)→  59Calculated saturated steam pressure(1852)→  60

Steam quality (1853)→  60Total mass flow (1854)→  60Condensate mass flow (1857)→  61

Energy flow (1872)→  61Heat flow difference (1863)→  61

Reynolds number (1864)→  62

Density (7607)→  62Pressure (7696)→  63

Saturation temperature (7709)→  63Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Compressibility factor (7729)→  64Vortex frequency (7722)→  64

Volume flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Volume flow (1838)

Description

Use this function to view the volume flow currently measured.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Volume flow unit parameter (→  70)

Corrected volume flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Correct.vol.flow (1850)

Description

Displays the corrected volume flow currently calculated.

User interface

Signed floating-point number

Additional information

Description
To calculate the corrected volume flow, the measured volume flow is multiplied by theratio of the density (Density parameter (→  62)) to the reference density. The density
and reference density here depend on the sensor version and the selected medium (seetable). Outputting the corrected volume flow cannot be used for gases that condense (e.g.
steam).
Sensor version

MediumMedium typeDensity

Reference density

Volume flow

All

1)

–

ρ

ρRef

Mass flow

Steam

–

f(p, T)–All except2)f(p, T)f(pRef, TRef)

LiquidAll except

2)

f(T)

f(TRef)Gas2)

f(p, T, z, pRef, TRef, zRef, ρRef)

ρRef

Gas

Endress+Hauser

57Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Sensor version

ρ
ρRef
p
pRef
T
TRef
z
zRef
alin
f(…)
1)
2)

MediumMedium typeDensity

Reference density

Liquid2)f(T, alin, TRef, ρRef)

ρRef

Fixed density (→  114)
Reference density (→  94)
Pressure (→  63)
Reference pressure (→  95)
Temperature (→  59)
Reference temperature (→  96)
Z-factor (→  94)
Reference Z-factor (→  96)
Linear expansion coefficient (→  91)
Calculation method as function of …
Outputting the corrected volume flow cannot be used for gases that condense.
User-specific gas or liquid

Dependency
The unit is taken from the Corrected volume flow unit parameter (→  73)

Mass flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Mass flow (1847)

Description

Displays the mass flow currently calculated.

User interface

Signed floating-point number

Additional information

Description
To calculate the mass flow, the measured volume flow is multiplied by the density(Density parameter (→  62)). The density depends on the sensor version and theselected medium (see table).
Sensor version

MediumMedium typeDensity

Volume flow

All

–

ρ

Mass flow

Steam–
All except

f(p, T)

LiquidAll except

1)

f(T)

Gas

1)

f(p, T, z, pRef, TRef, zRef, ρRef)

Liquid1)f(T, alin, TRef, ρRef)

Gas

ρ
ρRef
p
pRef
T
TRef
z
zRef
alin
f(…)
1)

58

f(p, T)
1)

Fixed density (→  114)
Reference density (→  94)
Pressure (→  63)
Reference pressure (→  95)
Temperature (→  59)
Reference temperature (→  96)
Z-factor (→  94)
Reference Z-factor (→  96)
Linear expansion coefficient (→  91)
Calculation method as function of …
User-specific gas or liquid

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Dependency
The unit is taken from the Mass flow unit parameter (→  72)

Flow velocity
Navigation

 Expert → Sensor → Measured val. → Process variab. → Flow velocity (1865)

Description

Displays the flow velocity currently calculated.

User interface

Signed floating-point number

Additional information

Description
The flow velocity is calculated based on the aspect ratio of the vortex measuring tube (DS)to the process connection of the sensor (DI), or to the Mating pipe diameter parameter(→  118) (DM) if this has been entered by the customer; the DS and DI are production
data that are defined by the shape and size of the meter body.
1

2

DM

DI

DS

A0025829

1
2
DM
DI
DS

Process connection
Vortex measuring tube
Mating pipe diameter (→  118)
Process connection of the sensor
Vortex measuring tube

Dependency
The unit is taken from the Velocity unit parameter (→  79)

Temperature
Navigation

 Expert → Sensor → Measured val. → Process variab. → Temperature (1851)

Description

Displays the temperature currently measured.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75).

Endress+Hauser

59Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Calculated saturated steam pressure
Navigation

 Expert → Sensor → Measured val. → Process variab. → CalcSatSteamPres (1852)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow"• In the Select medium parameter (→  85), the Steam option is selected.

Description

Displays the saturated steam pressure currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Pressure unit parameter (→  74)

Steam quality
Navigation

 Expert → Sensor → Measured val. → Process variab. → Steam quality (1853)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow"• In the Select medium parameter (→  85), the Steam option is selected.

Description

Displays the current steam quality. Depends on the compensation mode of the steam
quality (Steam quality parameter (→  116)).

User interface

Signed floating-point number

Total mass flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Total mass flow (1854)

Prerequisite

The following conditions are met:
• Order code for "Application package", option EU "Wet steam measurement"• In the Select medium parameter (→  85), the Steam option is selected.

Description

Displays the total mass flow (steam and condensate) currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Mass flow unit parameter (→  72)

60

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Condensate mass flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → CondensMassFlow (1857)

Prerequisite

The following conditions are met:
• Order code for "Application package", option EU "Wet steam measurement"• In the Select medium parameter (→  85), the Steam option is selected.

Description

Displays the condensate mass flow currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Mass flow unit parameter (→  72)

Energy flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Energy flow (1872)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Displays the energy flow currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Energy flow unit parameter (→  76)

Heat flow difference
Navigation

 Expert → Sensor → Measured val. → Process variab. → Heat flow diff. (1863)

Prerequisite

The following conditions are met:
Order code for "Sensor version", option "Mass flow"
One of the following options is selected in the Select gas type parameter (→  86):
• Single gas• Gas mixture• Natural gas• User-specific gas

Description

Displays the heat flow difference currently calculated.

User interface

Signed floating-point number

Endress+Hauser

61Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
The measuring device requires the following to calculate the heat flow difference correctly:
1.

Select the type of calculation in the Delta heat calculation parameter (→  114).

2.

Enter the value in the 2nd temperature delta heat parameter (→  115).

Dependency
The unit is taken from the Energy flow unit parameter (→  76)

Reynolds number
Navigation

 Expert → Sensor → Measured val. → Process variab. → Reynolds number (1864)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Displays the Reynolds number currently calculated.

User interface

Signed floating-point number

Additional information

Description
Re =

ρ⋅ν⋅d
η

Where:• ρ is the density of the medium (Density parameter (→  62))
• ν is the flow velocity of the fluid in relation to the body (Flow velocity parameter(→  59))• d is the characteristic length of the body• η is the viscosity of the medium– For gases: Dynamic viscosity parameter (→  93)– For liquids: Dynamic viscosity parameter (→  92)• The mating pipe diameter is taken as the characteristic length (Mating pipe diameterparameter (→  118))

Density
Navigation

 Expert → Sensor → Measured val. → Process variab. → Density (7607)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Displays the density currently calculated.

User interface

Positive floating-point number

62

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
Depending on the selected medium the density is calculated with pressure andtemperature and the corresponding method (e.g. IAPWS, NEL40...).
Dependency
The unit is taken from the Density unit parameter (→  79)

Pressure
Navigation

 Expert → Sensor → Measured val. → Process variab. → Pressure (7696)

Prerequisite

For the following order code:• "Sensor version", option "Mass flow"• In the External value parameter (→  113), the Pressure option is selected.

Description

Displays the current process pressure.

User interface

0 to 250 bar

Additional information

Description
The value of the pressure which is read in (e.g. via the current input module) is displayed..
If the Pressure option is not selected as the external value in the External value
parameter (→  113), the input value for the fixed process pressure (Fixed processpressure parameter (→  115)) is displayed.
Dependency
The unit is taken from the Pressure unit parameter (→  74)

Saturation temperature
Navigation

 Expert → Sensor → Measured val. → Process variab. → Saturation temp. (7709)

Prerequisite

In the Select medium parameter (→  85), the Steam option is selected.

Description

Displays the saturation temperature currently calculated.

User interface

Country-specific:
• °C
• °F

Additional information

The saturation temperature describes the temperature limit at which steam begins tocondense. This value is calculated using the current process pressure (Pressure parameter(→  63)) according to IAPWS-IF97.
Dependency
The unit is taken from the Temperature unit parameter (→  75).

Endress+Hauser

63Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Compressibility factor
Navigation

 Expert → Sensor → Measured val. → Process variab. → CompressFactor (7729)

Prerequisite

The following conditions are met:
Order code for "Sensor version", option "Mass flow"
In the Select medium parameter (→  85), the Gas option or Steam option is selected.

Description

Displays the compressibility factor currently calculated.

User interface

0 to 2

Additional information

Description
The compressibility factor describes the deviation of the medium from the ideal behavior
under the current process conditions. If the medium is a user-specific gas/liquid, thecompressibility factor is entered as the Z-factor (Z-factor parameter (→  94)).

Vortex frequency
Navigation

 Expert → Sensor → Measured val. → Process variab. → Vortex frequency (7722)

Description

Displays the measured variable for the flow in the measuring tube which is recorded
directly with the DSC sensor.

User interface

Measuring range depending on the nominal diameter:
0.1 to 3 100 Hz

Additional information

Description
The filter settings specify the measuring range of the vortex frequency depending on thenominal diameter.
Filter settings for liquids
Nominal diameter

64

Minimum vortex frequency

Maximum vortex frequency

fvmin
[Hz]

fvmax
[Hz]

DN 15 (½")
DN 25 (1") > DN 15 (½")
DN 40 (1½") >> DN 15 (½")

11.5

666.5

DN 25 (1")
DN 40 (1½") > DN 25 (1")
DN 50 (2") >> DN 25 (1")

6.7

388.8

DN 40 (1½")
DN 50 (2") > DN 40 (1½")
DN 80 (3") >> DN 40 (1½")

3.9

224.3

DN 50 (2")
DN 80 (3") > DN 50 (2")
DN 100 (4") >> DN 50 (2")

3.0

172.8

DN 80 (3")
DN 100 (4") > DN 80 (3")
DN150 (6") >> DN 80 (3")

2.1

122.8

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Nominal diameter

Description of device parameters

Minimum vortex frequency

Maximum vortex frequency

fvmin
[Hz]

fvmax
[Hz]

DN 100 (4")
DN150 (6") > DN 100 (4")
DN 200 (8") >> DN 100 (4")

1.7

101.4

DN150 (6")
DN 200 (8") > DN150 (6")
DN 250 (10") >> DN150 (6")

1.1

66.6

DN 200 (8")

0.7

41.7

DN 250 (10")

0.6

34.3

DN 300 (12")

0.5

28.9

Minimum vortex frequency

Maximum vortex frequency

fvmin
[Hz]

fvmax
[Hz]

DN 15 (½")
DN 25 (1") > DN 15 (½")
DN 40 (1½") >> DN 15 (½")

209.9

3 100

DN 25 (1")
DN 40 (1½") > DN 25 (1")
DN 50 (2") >> DN 25 (1")

67.1

3 100

DN 40 (1½")
DN 50 (2") > DN 40 (1½")
DN 80 (3") >> DN 40 (1½")

13.7

1 869.1

DN 50 (2")
DN 80 (3") > DN 50 (2")
DN 100 (4") >> DN 50 (2")

10.5

2 303.8

DN 80 (3")
DN 100 (4") > DN 80 (3")
DN150 (6") >> DN 80 (3")

7.5

1 636.9

DN 100 (4")
DN150 (6") > DN 100 (4")
DN 200 (8") >> DN 100 (4")

6.2

1 352.3

DN150 (6")
DN 200 (8") > DN150 (6")
DN 250 (10") >> DN150 (6")

4.1

888.6

DN 200 (8")

2.5

555.4

DN 250 (10")

2.1

457.3

DN 300 (12")

1.8

385.3

Filter settings for gases/steam
DN[mm (in)]Endress+Hauser

65Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Totalizer
Navigation

 Expert → Sensor → Measured val. → Totalizer‣ Totalizer
Totalizer value 1 to 3 (0911–1 to 3)→  66

Totalizer overflow 1 to 3 (0910–1 to 3)→  67

Totalizer value 1 to 3

Navigation

 Expert → Sensor → Measured val. → Totalizer → Totalizer val. 1 to 3 (0911–1 to 3)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  257) of the Totalizer 1 to 3 submenu:• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *Description

Displays the current totalizer reading.

User interface

Signed floating-point number

Additional information

Description
As it is only possible to display a maximum of 7 digits, the current counter value is the sumof the totalizer value and the overflow value from the Totalizer overflow 1 to 3 parameter
if the display range is exceeded.
In the event of an error, the totalizer adopts the mode defined in the Failure modeparameter (→  261).
User interface
The unit of the selected process variable is specified for the totalizer in the Unit
totalizer parameter (→  258).
ExampleCalculation of the current totalizer reading when the value exceeds the 7-digit display
range:• Value in the Totalizer value 1 parameter: 196 845.7 m3• Value in the Totalizer overflow 1 parameter: 1 ⋅ 106 (1 overflow) = 1 000 000 [m3]• Current totalizer reading: 1 196 845.7 m3*

66

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Totalizer overflow 1 to 3

Navigation

 Expert → Sensor → Measured val. → Totalizer → Tot. overflow 1 to 3 (0910–1 to 3)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  257) of the Totalizer 1 to 3 submenu:• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *Description

Displays the current totalizer overflow.

User interface

Integer with sign

Additional information

Description
If the current reading has more than 7 digits, which is the maximum value range that canbe displayed, the value above this range is given as an overflow. The current totalizer value
is therefore the sum of the overflow value and the totalizer value from the Totalizer value
1 to 3 parameter.
User interface
The unit of the selected process variable is specified for the totalizer in the Unit
totalizer parameter (→  258).
ExampleCalculation of the current totalizer reading when the value exceeds the 7-digit display
range:• Value in the Totalizer value 1 parameter: 196 845.7 m3• Value in the Totalizer overflow 1 parameter: 2 ⋅ 106 (2 overflows) = 2 000 000 [m3]• Current totalizer reading: 2 196 845.7 m3"Output values" submenu
Navigation

 Expert → Sensor → Measured val. → Output values‣ Output values*

Terminal voltage 1 (0662)→  68

Pulse output (0456)→  68Visibility depends on order options or device settings

Endress+Hauser

67Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Output frequency (0471)→  69Switch status (0461)→  69Terminal voltage 1
Navigation

 Expert → Sensor → Measured val. → Output values → Terminal volt. 1 (0662)

Description

Use this function to view the actual terminal voltage that is present at the current output.

User interface

0.0 to 50.0 V

Pulse output
Navigation

 Expert → Sensor → Measured val. → Output values → Pulse output (0456)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  121).

Description

Displays the pulse frequency currently output.

User interface

Positive floating-point number

Additional information

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.
• The Value per pulse parameter (→  124) and the Pulse width parameter(→  124) can be used to define the value (i.e. the amount of the measured value that
corresponds to a pulse) and the duration of the pulse.
1
NC
0

1
NO
0
A0025816-EN

0
1
NC
NO

68

Non-conductive
Conductive
NC contact (normally closed)
NO contact (normally open)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

The output behavior can be reversed via the Invert output signal parameter (→  138),
i.e. the transistor does not conduct for the duration of the pulse.
In addition, the behavior of the output in the event of an error (Failure mode parameter(→  125)) can be configured.

Output frequency
Navigation

 Expert → Sensor → Measured val. → Output values → Output freq. (0471)

Prerequisite

In the Operating mode parameter (→  121), the Frequency option is selected.

Description

Displays the actual value of the output frequency which is currently measured.

User interface

0 to 1 250 Hz

Switch status
Navigation

 Expert → Sensor → Measured val. → Output values → Switch status (0461)

Prerequisite

The Switch option is selected in the Operating mode parameter (→  121).

Description

Displays the current switch status of the status output.

User interface• Open• Closed

Additional information

User interface• OpenThe switch output is not conductive.
• Closed
The switch output is conductive.

3.2.2"System units" submenu

Navigation

 Expert → Sensor → System units‣ System units

Endress+Hauser

Volume flow unit (0553)→  70Volume unit (0563)→  72Mass flow unit (0554)→  7269

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Mass unit (0574)→  73

Corrected volume flow unit (0558)→  73

Corrected volume unit (0575)→  74Pressure unit (0564)→  74Temperature unit (0557)→  75

Energy flow unit (0565)→  76

Energy unit (0559)→  77

Calorific value unit (0552)→  77

Calorific value unit (0606)→  78Velocity unit (0566)→  79Density unit (0555)→  79Dynamic viscosity unit (0577)→  80Specific heat capacity unit (0604)→  80Length unit (0551)→  81

Date/time format (2812)→  81

Volume flow unitNavigation

 Expert → Sensor → System units → Volume flow unit (0553)

Description

Use this function to select the unit for the volume flow.

70

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dFactory setting

Country-specific:• m³/h• ft³/min

Additional information

ResultDescription of device parameters

US units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)The selected unit applies for:Volume flow parameter (→  57)
Selection
For an explanation of the abbreviated units: →  298

Endress+Hauser

71Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Volume unitNavigation

 Expert → Sensor → System units → Volume unit (0563)

Description

Use this function to select the unit for the volume.

Selection

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega

Factory setting

Country-specific:
• m³
• ft³

Additional information

Selection

US units• af• ft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;beer)• bbl (us;tank)

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

For an explanation of the abbreviated units: →  298

Mass flow unitNavigation

 Expert → Sensor → System units → Mass flow unit (0554)

Description

Use this function to select the unit for the mass flow.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

Factory setting

Country-specific:• kg/h• lb/min72

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/dEndress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

ResultThe selected unit applies for:• Mass flow parameter (→  58)• Total mass flow parameter (→  60)• Condensate mass flow parameter (→  61)
Selection
For an explanation of the abbreviated units: →  298

Mass unitNavigation

 Expert → Sensor → System units → Mass unit (0574)

Description

Use this function to select the unit for the mass.

Selection

SI units
•g
• kg
•t

Factory setting

Country-specific:
• kg
• lb

Additional information

Selection

US units• oz• lb• STonFor an explanation of the abbreviated units: →  298

Corrected volume flow unitNavigation

 Expert → Sensor → System units → Cor.volflow unit (0558)

Description

Use this function to select the unit for the corrected volume flow.

Selection

SI units• Nl/s• Nl/min• Nl/h• Nl/d• Nm³/s• Nm³/min• Nm³/h• Nm³/d• Sm³/s• Sm³/min• Sm³/h• Sm³/dEndress+Hauser

US units• Sft³/s• Sft³/min• Sft³/h• Sft³/d73

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

Country-specific:• Nm³/h• Sft³/h

Additional information

ResultThe selected unit applies for:
Corrected volume flow parameter (→  57)
Selection
For an explanation of the abbreviated units: →  298

Corrected volume unitNavigation

 Expert → Sensor → System units → Corr. vol. unit (0575)

Description

Use this function to select the unit for the corrected volume.

Selection

SI units
• Nl
• Nm³
• Sm³

Factory setting

Country-specific:• Nm³• Sft³

Additional information

Selection

US units
Sft³

For an explanation of the abbreviated units: →  298

Pressure unitNavigation

 Expert → Sensor → System units → Pressure unit (0564)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to select the unit for the pipe pressure.

74

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

SI units
• Pa
• kPa
• MPa
• mbar a• bar• torr• atm• gf/cm²• kgf/cm²

Factory setting

Country-specific:
• bar
• psi

Additional information

ResultDescription of device parameters

US unitspsi

Other units• mmH2O (4°C)• mmH2O (68°F)• mmHg (0°C)• inH2O (4°C)• inH2O (68°F)• ftH2O (68°F)• inHg (0°C)

The unit is taken from:• Calculated saturated steam pressure parameter (→  60)• Atmospheric pressure parameter (→  113)
• Maximum value parameter (→  287)
• Fixed process pressure parameter (→  115)
• Pressure parameter (→  63)• Reference pressure parameter (→  95)
Selection
For an explanation of the abbreviated units: →  298

Temperature unitNavigation

 Expert → Sensor → System units → Temperature unit (0557)

Description

Use this function to select the unit for the temperature.

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

ResultUS units• °F• °R

The selected unit applies for:• Temperature parameter (→  59)• Maximum value parameter (→  284)
• Minimum value parameter (→  284)
• Average value parameter (→  284)
• Maximum value parameter (→  285)
• Minimum value parameter (→  285)
• Maximum value parameter (→  286)
• Minimum value parameter (→  286)
Endress+Hauser

75Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• 2nd temperature delta heat parameter (→  115)
• Fixed temperature parameter (→  114)
• Reference combustion temperature parameter (→  94)• Reference temperature parameter (→  96)• Saturation temperature parameter (→  63)
Selection
For an explanation of the abbreviated units: →  298

Energy flow unitNavigation

 Expert → Sensor → System units → Energy flow unit (0565)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to select the unit for the energy flow.

Selection

SI units
• kW
• MW
• GW
• kJ/s• kJ/min• kJ/h• kJ/d• MJ/s• MJ/h• MJ/min• MJ/d• GJ/s• GJ/min• GJ/h• GJ/d• kcal/s• kcal/min• kcal/h• kcal/d• Mcal/s• Mcal/min• Mcal/h• Mcal/d• Gcal/s• Gcal/min• Gcal/h• Gcal/d

Factory setting

Country-specific:• kW• Btu/h76

Imperial units• Btu/s• Btu/min• Btu/h• Btu/day• MBtu/s• MBtu/min• MBtu/h• MBtu/d• MMBtu/s• MMBtu/min• MMBtu/h• MMBtu/dEndress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

ResultThe selected unit applies for:• Heat flow difference parameter (→  61)• Energy flow parameter (→  61)
Selection
For an explanation of the abbreviated units: →  298

Energy unitNavigation

 Expert → Sensor → System units → Energy unit (0559)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to select the unit for energy.

Selection

SI units
• kWh
• MWh
• GWh
• kJ
• MJ
• GJ
• kcal• Mcal• GcalFactory setting

Country-specific:
• kWh
• Btu

Additional information

Selection

Imperial units• Btu• MBtu• MMBtu

For an explanation of the abbreviated units: →  298

Calorific value unitNavigation

 Expert → Sensor → System units → Cal. value unit (0552)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow"• The Gross calorific value volume option or the Net calorific value volume option is
selected in the Calorific value type parameter (→  90).

Description

Use this function to select the unit for the calorific value.

Endress+Hauser

77Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

SI units• kJ/Nm³• MJ/Nm³• kWh/Nm³• MWh/Sm³• kJ/Sm³• MJ/Sm³• kWh/Sm³• MWh/Nm³

Factory setting

Country-specific:• kJ/Nm³• Btu/Sft³

Additional information

ResultImperial units• Btu/Sm³• MBtu/Sm³• Btu/Sft³• MBtu/Sft³

The selected unit applies for:
Reference gross calorific value parameter (→  95)
Selection
For an explanation of the abbreviated units: →  298

Calorific value unit (Mass)

Navigation

 Expert → Sensor → System units → Cal. value unit (0606)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow"• The Gross calorific value mass option or the Net calorific value mass option is selectedin the Calorific value type parameter (→  90).

Description

Use this function to select the unit for the calorific value (mass).

Selection

SI units• kJ/kg• MJ/kg• kWh/kg• MWh/kg

Factory setting

Country-specific:• kJ/kg• Btu/lb

Additional information

Selection

US units• kJ/lb• MJ/lb• kWh/lb• MWh/lb

Imperial units• Btu/lb• MBtu/lb

For an explanation of the abbreviated units: →  298

78

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Velocity unitNavigation

 Expert → Sensor → System units → Velocity unit (0566)

Description

Use this function to select the unit for the flow velocity.

Selection

SI units
m/s

Factory setting

Country-specific:• m/s• ft/sAdditional information

ResultUS units
ft/s

The selected unit applies for:• Flow velocity parameter (→  59)• Maximum value parameter (→  287)
Selection
For an explanation of the abbreviated units: →  298

Density unitNavigation

 Expert → Sensor → System units → Density unit (0555)

Description

Use this function to select the unit for the density.

Selection

SI units• g/cm³• kg/dm³• kg/l• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

Factory setting

Country-specific:• kg/m³• lb/ft³

Additional information

ResultUS units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)The selected unit applies for:• Density parameter (→  62)• Fixed density parameter (→  114)
• Reference density parameter (→  94)

Endress+Hauser

79Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• SD = specific density
The specific density is the ratio of the fluid density to the water density at a watertemperature of +4 °C (+39 °F), +15 °C (+59 °F), +20 °C (+68 °F).
• SG = specific gravity
The specific gravity is the ratio of the fluid density to the water density at a watertemperature of +4 °C (+39 °F), +15 °C (+59 °F), +20 °C (+68 °F).
For an explanation of the abbreviated units: →  298

Dynamic viscosity unitNavigation

 Expert → Sensor → System units → Dyn. visc. unit (0577)

Description

Use this function to select the unit for dynamic viscosity.

Selection

SI units• Pa s• cP•P

Factory setting

Pa s

Additional information

ResultThe selected unit applies for:• Dynamic viscosity parameter (→  93) (gases)• Dynamic viscosity parameter (→  92) (liquids)

Additional information

Selection
For an explanation of the abbreviated units: →  298

Specific heat capacity unitNavigation

 Expert → Sensor → System units → SpecHeatCapaUnit (0604)

Prerequisite

The following conditions are met:
• Selected medium:– The User-specific gas option is selected in the Select gas type parameter(→  86).
Or– The User-specific liquid option is selected in the Select liquid type parameter(→  87).
• The Heat option is selected in the Enthalpy type parameter (→  90).

Description

Use this function to select the unit for the specific heat capacity.

80

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

SI units• kJ/(kgK)• MJ/(kgK)• kWh/(kgK)• kcal/(kgK)Factory setting

kJ/(kgK)

Additional information

ResultDescription of device parameters

Imperial units
Btu/(lb°R)

The selected unit applies for:Specific heat capacity parameter (→  97)
Additional information

Selection
For an explanation of the abbreviated units: →  298

Length unitNavigation

 Expert → Sensor → System units → Length unit (0551)

Description

Use this function to select the unit of length for the nominal diameter.

Selection

SI units• mm•m

Factory setting

Country-specific:
• mm
• in

Additional information

ResultUS units• in• ft

The selected unit applies for:• Inlet run parameter (→  117)• Mating pipe diameter parameter (→  118)
Selection
For an explanation of the abbreviated units: →  298

Date/time format



Navigation

 Expert → Sensor → System units → Date/time format (2812)

Description

Use this function to select the desired time format for calibration history.

Endress+Hauser

81Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• dd.mm.yy hh:mm• dd.mm.yy hh:mm am/pm• mm/dd/yy hh:mm• mm/dd/yy hh:mm am/pm

Factory setting

dd.mm.yy hh:mm

Additional information

Selection
For an explanation of the abbreviated units: →  298

3.2.3
Navigation"Process parameters" submenu
 Expert → Sensor → Process param.

‣ Process parameters
Flow override (1839)→  82

Flow damping (1802)→  83‣ Low flow cut off→  83

Flow override

Navigation

 Expert → Sensor → Process param. → Flow override (1839)

Description

Use this function to select whether to interrupt the evaluation of measured values. This isuseful for the cleaning process of a pipeline, for example.

Selection

• Off
• On

Factory setting

Off

Additional information

ResultThis setting affects all the functions and outputs of the measuring device.
Description
Flow override is active• The diagnostic message diagnostic message C453 Flow override is displayed.
• Output values– Output: Value at zero flow
– Temperature: Proceeding output– Totalizers 1-3: Stop being totalized

82

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Flow damping

Navigation

 Expert → Sensor → Process param. → Flow damping (1802)

Description

Use this function to enter flow damping. Reduction of the variability of the flow measured
value (in relation to interference). For this purpose, the depth of the flow filter is adjusted:when the filter setting increases, the reaction time of the device also increases.

User entry

0 to 999.9 s

Factory setting

5sAdditional information

Result
The damping has an effect on the following variables of the device:• Outputs• Low flow cut off →  83
• Totalizer →  256
User entry• Value = 0: no damping• Value > 0: damping is increased"Low flow cut off" submenu
Navigation

 Expert → Sensor → Process param. → Low flow cut off‣ Low flow cut off
Assign process variable (1837)→  83On value low flow cutoff (1805)→  84

Off value low flow cutoff (1804)→  84

Assign process variable

Navigation

 Expert → Sensor → Process param. → Low flow cut off → Assign variable (1837)

Description

Use this function to select a process variable for low flow cut off.

Endress+Hauser

83Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Off• Volume flow
• Corrected volume flow
• Mass flow
• Reynolds number *Factory settingOff

On value low flow cutoff

Navigation

 Expert → Sensor → Process param. → Low flow cut off → On value (1805)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  83):• Volume flow
• Corrected volume flow
• Mass flow
• Reynolds number *Description

Use this function to enter a switch-on value for low flow cut off. Low flow cut off isactivated if the value entered is not equal to 0 →  84.

User entry

Positive floating-point number

Factory setting0

Additional information

Dependency
The unit depends on the process variable selected in the Assign process variable
parameter (→  83).

Off value low flow cutoff

Navigation

 Expert → Sensor → Process param. → Low flow cut off → Off value (1804)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  83):• Volume flow
• Corrected volume flow
• Mass flow
• Reynolds number *Description

Use this function to enter a switch-off value for low flow cut off. The off value is entered asa positive hysteresis from the on value →  84.

User entry

0 to 100.0 %Factory setting

50 %

*

84

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Example
Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn value entered
Off value entered

3.2.4
Navigation"Measurement mode" submenu
 Expert → Sensor → Measurement mode

‣ Measurement mode
Select medium (7653)→  85

Select gas type (7635)→  86

Select liquid type (7636)→  87

Select steam type (7637)→  87

Density calculation (7608)→  88Enthalpy calculation (7619)→  88Saturated steam calculation mode
(7708)→  89‣ Medium properties→  89Select medium



Navigation

 Expert → Sensor → Measurement mode → Select medium (7653)

Description

Use this function to select the type of medium for the measuring application.

Endress+Hauser

85Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Gas• Liquid• Steam

Factory setting

Steam

Select gas type

Navigation

 Expert → Sensor → Measurement mode → Select gas type (7635)

Prerequisite

The following conditions are met:
• Order code
– "Sensor version", option "Mass flow"– "Application package", option "Air + Industrial gases" or option "Natural gas"
• The Gas option is selected in the Select medium parameter (→  85).

Description

Use this function to select the type of gas for the measuring application.

Selection• Single gas• Gas mixture• Air• Natural gas• User-specific gas

Factory settingUser-specific gas

Additional information"User-specific gas" option
Applications: calculation of the mass flow of a user-specific gas
Calculated variables: the mass flow, the density, the corrected volume flow and the heat
quantity are calculated from the measured volume flow and the measured temperature.
Either the specific thermal capacity or the calorific value must be entered for calculating
the heat quantity.
Formulae for calculation:• Mass flow: m = q ⋅ ρ (T)• Density: ρ = ρ1 (T1) / (1 + βp ⋅ [T - T1])
• Corrected volume flow: vn = q ⋅ (ρ (T) / ρref)• Heat quantity in the case of delta heat: E = q ⋅ ρ (T) ⋅ cp ⋅ ΔT
• Heat quantity in the case of combustion: E = q ⋅ ρ (T) ⋅ h
m = Mass flowq = Volume flow (measured)vn = Corrected volume flow
T = Process temperature (measured)T1 = Temperature (→  59) at which the value for ρ1 applies.
ρ = Density
ρref = Reference density= Density (→  62) at which the value for T1 applies.
βp = Linear expansion coefficient (→  91) of the liquid at T1
Possible combinations of these values: Linear expansion coefficient parameter(→  91)

86

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Select liquid type

Navigation

 Expert → Sensor → Measurement mode → Sel. liquid type (7636)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow"• The Liquid option is selected in the Select medium parameter (→  85) parameter.

Description

Use this function to select the type of liquid for the measuring application.

Selection• Water• LPG (Liquefied Petroleum Gas)
• User-specific liquid

Factory setting

Water

Additional information"User-specific liquid" option
Applications: calculation of the mass flow of a user-specific liquid, such as thermal oil.
Calculated variables: the mass flow, the density, the corrected volume flow and the heat
quantity are calculated from the measured volume flow and the measured temperature.
Either the specific thermal capacity or the calorific value must be entered for calculating
the heat quantity.
Formulae for calculation:• Mass flow: m = q ⋅ ρ (T)• Density: ρ = ρ1 (T1) / (1 + βp ⋅ [T - T1])
• Corrected volume flow: vn = q ⋅ (ρ (T) / ρref)• Heat quantity in the case of delta heat: E = q ⋅ ρ (T) ⋅ cp ⋅ ΔT
• Heat quantity in the case of combustion: E = q ⋅ ρ (T) ⋅ h
m = Mass flowq = Volume flow (measured)vn = Corrected volume flow
T = Process temperature (measured)T1 = Temperature (→  59) at which the value for ρ1 applies.
ρ = Density
ρref = Reference density= Density (→  62) at which the value for T1 applies.
βp = Linear expansion coefficient (→  91) of the liquid at T1
Possible combinations of these values: Linear expansion coefficient parameter(→  91)

Select steam type

Navigation

 Expert → Sensor → Measurement mode → Sel. steam type (7637)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow (integrated temperature
measurement)"• In the Select medium parameter (→  85), the Steam option is selected.

Endress+Hauser

87Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Description

Use this function to select the type of steam for the measuring application.

Selection• Wet steam• Superheated steam• Saturated steam

Factory setting

Saturated steam

Density calculation



Navigation

 Expert → Sensor → Measurement mode → Density calc. (7608)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.

Description

Use this function to select the standard on the basis of which the density is calculated.

Selection• AGA Nx19
• ISO 12213- 2• ISO 12213- 3

Factory setting

AGA Nx19

Enthalpy calculation



Navigation

 Expert → Sensor → Measurement mode → Enthalpy calc. (7619)

Prerequisite

The following conditions are met:
• Order code
– "Sensor version", option "Mass flow (integrated temperature measurement)"– "Application package", option "Natural gas"
• In the Select medium parameter (→  85), the Gas option is selected and in the Selectgas type parameter (→  86), the Natural gas option is selected.

Description

Use this function to select the standard on the basis of which the enthalpy is calculated.

Selection• AGA5• ISO 6976

Factory setting

AGA5

88

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Saturated steam calculation modeNavigation

 Expert → Sensor → Measurement mode → SatSteamCalcMode (7708)

Prerequisite

In the Select steam type parameter (→  87), the Wet steam option or Saturated steam
option is selected.

Description

Use this function to select the calculation method for the saturated steam application.

Selection• Temperature• Pressure

Factory setting

Temperature"Medium properties" submenu
Navigation

 Expert → Sensor → Measurement mode → Medium property‣ Medium properties

Endress+Hauser

Enthalpy type (7620)→  90Calorific value type (7698)→  90

Reference combustion temperature(7699)→  94

Reference density (7700)→  94

Reference gross calorific value (7701)→  95

Reference pressure (7702)→  95

Reference temperature (7703)→  96

Reference Z-factor (7704)→  96

Linear expansion coefficient (7621)→  91

Relative density (7705)→  96

Specific heat capacity (7716)→  97

Calorific value (7626)→  92Z-factor (7631)→  9489

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Dynamic viscosity (7733)→  92

Dynamic viscosity (7732)→  93‣ Gas composition→  97

Calorific value type

Navigation

 Expert → Sensor → Measurement mode → Medium property → Cal. value type(7698)

Prerequisite

The Calorific value type parameter (→  90) is visible.

Description

Use this function to select whether the net calorific value or the gross calorific value isused as the basis for calculation.

Selection• Gross calorific value volume• Net calorific value volume• Gross calorific value mass
• Net calorific value mass

Factory setting

Gross calorific value mass

Enthalpy type

Navigation

 Expert → Sensor → Measurement mode → Medium property → Enthalpy type(7620)

Prerequisite

The following conditions are met:
• In the Select gas type parameter (→  86), the User-specific gas option is selected.
Or• In the Select liquid type parameter (→  87), the User-specific liquid option is
selected.

Description

Use this function to select the type of enthalpy.

Selection• Heat• Calorific value

Factory setting

Heat

90

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Linear expansion coefficientNavigation

 Expert → Sensor → Measurement mode → Medium property → Linear exp coeff(7621)

Prerequisite

The following conditions are met:
• The Liquid option is selected in the Select medium parameter (→  85).
• The User-specific liquid option is selected in the Select liquid type parameter(→  87).

Description

Use this function to enter the linear, medium-specific expansion coefficient for calculating
the reference density for user-specific liquids.

User entry

1.0-6 to 2.0-3

Factory setting

2.06-4

Additional information

User entry• If the value in this parameter is changed, it is advisable to reset the totalizer.
• The expansion coefficient can be determined using the Applicator.
• If two density and temperature value pairs are known (density ρ1 at temperature T1 anddensity ρ2 at temperature T2), the expansion coefficient can be calculated according to
the following formula:βp = ((ρ1/ρ2) - 1)/(T1 - +T2)
Sample values
The closer the process temperature is to the specific temperature value, the better the
calculation of the density for application-specific liquids. If the process temperaturedeviates greatly from the value indicated, the expansion coefficient should be
calculated according to the formula (see above).

Endress+Hauser

Medium(liquid)

Temperature value[K]

Density value[kg/m³]

Expansion coefficient[10-4 1/K]

Air

123.15594

18.76

Ammonia298.15

602

25

Argon

133.15

1 028

111.3

n-butane

298.15573

20.7

Carbon dioxide298.15713

106.6

Chlorine

298.15

1 398

21.9

Cyclohexane

298.15773

11.6

n-decane

298.15728

10.2

Ethane298.15315

175.3

Ethylene

298.15386

87.7

n-heptane298.15351

12.4

n-hexane

298.15656

13.8

Hydrogen chloride

298.15796

70.9

i-butane

298.15552

22.5

Methane163.15331

73.5

Nitrogen

93.15

729

75.3

n-octane

298.15

699

11.1

91

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Medium(liquid)

Temperature value[K]

Density value[kg/m³]

Expansion coefficient[10-4 1/K]

Oxygen133.15876

95.4

n-pentane298.15621

16.2

Propane298.15493

32.1

Vinyl chloride

298.15903

19.3

Table values according to Carl L. Yaws (2001): Matheson Gas Data Book, 7th edition

Calorific value

Navigation

 Expert → Sensor → Measurement mode → Medium property → Calorific value(7626)

Prerequisite

The following conditions are met:
• Selected medium:– In the Select gas type parameter (→  86), the User-specific gas option is selected.
Or– In the Select liquid type parameter (→  87), the User-specific liquid option is
selected.
• In the Enthalpy type parameter (→  90), the Calorific value option is selected.
• In the Calorific value type parameter (→  90), the Gross calorific value volume
option or Gross calorific value mass option is selected.

Description

Use this function to enter the calorific value for calculating the energy flow.

User entry

Positive floating-point number

Factory setting

50 000 kJ/kg

Dynamic viscosity (Liquids)

Navigation

 Expert → Sensor → Measurement mode → Medium property → Dynam. viscosity(7733)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Volume flow"• The Liquid option is selected in the Select medium parameter (→  85) parameter.
Or• The User-specific liquid option is selected in the Select liquid type parameter(→  87).

Description

Use this function to enter a fixed value for the dynamic viscosity for a liquid.

User entry

Positive floating-point number

Factory setting

1 cP

92

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
The viscosity entered is used to linearize the measured error in the lower Reynolds number
range if the calculated viscosity is not available e.g. "Volume flow" sensor version or the
fluid is a user-specific liquid (see table).
Dependencies
Sensor version

MediumVolume flow

All

Mass flow

All except

Dyn. viscosity
x
1)

1)

x
1)

–
x

Dynamic viscosity as the input value
User-specific liquid

Dependency
The unit is taken from the Dynamic viscosity unit parameter (→  80)

Dynamic viscosity (Gases)

Navigation

 Expert → Sensor → Measurement mode → Medium property → Dynam. viscosity(7732)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Volume flow"• The Gas option or the Steam option is selected in the Select medium parameter(→  85).
Or• The User-specific gas option is selected in the Select gas type parameter (→  86).

Description

Use this function to enter a fixed value for the dynamic viscosity for a gas or steam.

User entry

Positive floating-point number

Factory setting

0.015 cP

Additional information

Description
The viscosity entered is used to linearize the measured error in the lower Reynolds number
range if the calculated viscosity is not available e.g. "Volume flow" sensor version or the
fluid is a user-specific gas (see table).
Dependencies
Sensor version

MediumDyn. viscosity

Volume flowAllx

Mass flow

All except 1)

–

1)

x

x
1)

Endress+Hauser

Dynamic viscosity as the input value
User-specific gas

93

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Dependency
The unit is taken from the Dynamic viscosity unit parameter (→  80)

Z-factor



Navigation

 Expert → Sensor → Measurement mode → Medium property → Z-factor (7631)

Prerequisite

In the Select gas type parameter (→  86), the User-specific gas option is selected.

Description

Use this function to enter the real gas constant Z for gas under operating conditions.

User entry

0.1 to 2.0

Factory setting

1

Reference combustion temperature

Navigation

 Expert → Sensor → Measurement mode → Medium property → Ref. comb. temp.
(7699)

Prerequisite

The Reference combustion temperature parameter (→  94) is visible.

Description

Use this function to enter the reference combustion temperature for calculating thenatural gas energy value.

User entry–200 to 450 °C

Factory setting

20 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75)

Reference density

Navigation

 Expert → Sensor → Measurement mode → Medium property → Ref.density (7700)

Prerequisite

The following conditions are met:
• In the Select gas type parameter (→  86), the User-specific gas option is selected.
Or• In the Select liquid type parameter (→  87), the Water option or User-specific liquid
option is selected.

Description

Use this function to enter a fixed value for the reference density.

94

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User entry

0.01 to 15 000 kg/m³

Factory setting

1 000 kg/m³

Additional information

Dependency

Description of device parameters

The unit is taken from the Density unit parameter (→  79)

Reference gross calorific value

Navigation

 Expert → Sensor → Measurement mode → Medium property → Ref. GrossCalVal(7701)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 3 option is selected.

Description

Use this function to enter the reference gross calorific value of the natural gas.

User entry

Positive floating-point number

Factory setting

50 000 kJ/Nm³

Additional information

Dependency
The unit is taken from the Calorific value unit parameter (→  77)

Reference pressureNavigation

 Expert → Sensor → Measurement mode → Medium property → Ref. pressure (7702)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow (integrated temperature
measurement)"• The Gas option is selected in the Select medium parameter (→  85).

Description

Use this function to enter the reference pressure for calculating the reference density.

User entry

0 to 250 bar

Factory setting

1.01325 bar

Additional information

Dependency
The unit is taken from the Pressure unit parameter (→  74)

Endress+Hauser

95Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Reference temperature

Navigation

 Expert → Sensor → Measurement mode → Medium property → Ref. temperature(7703)

Prerequisite

The following conditions are met:
• The Gas option is selected in the Select medium parameter (→  85).
Or
• The Liquid option is selected in the Select medium parameter (→  85).

Description

Use this function to enter the reference temperature for calculating the reference density.

User entry–200 to 450 °C

Factory setting

20 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75)

Reference Z-factor



Navigation

 Expert → Sensor → Measurement mode → Medium property → Ref. Z-factor (7704)

Prerequisite

In the Select gas type parameter (→  86), the User-specific gas option is selected.

Description

Use this function to enter the real gas constant Z for gas under reference conditions.

User entry

0.1 to 2

Factory setting1

Relative density

Navigation

 Expert → Sensor → Measurement mode → Medium property → Relative density(7705)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 3 option is selected.

Description

Use this function to enter the relative density of the natural gas.

User entry

0.55 to 0.9

Factory setting

0.664

96

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Specific heat capacity



Navigation

 Expert → Sensor → Measurement mode → Medium property → Spec. heat cap.
(7716)

Prerequisite

The following conditions are met:
• Selected medium:– In the Select gas type parameter (→  86), the User-specific gas option is selected.
Or– In the Select liquid type parameter (→  87), the User-specific liquid option is
selected.
• In the Enthalpy type parameter (→  90), the Heat option is selected.

Description

Use this function to enter the specific heat capacity of the medium.

User entry

0 to 50 kJ/(kgK)Factory setting

4.187 kJ/(kgK)

Additional information

Dependency
The unit is taken from the Specific heat capacity unit parameter (→  80)"Gas composition" submenu
Navigation

 Expert → Sensor → Measurement mode → Medium property→ Gas composition‣ Gas composition

Endress+Hauser

Gas type (7714)→  99Gas mixture (7640)→  99Mol% Ar (7663)→  100

Mol% C2H3Cl (7664)→  100

Mol% C2H4 (7665)→  101

Mol% C2H6 (7666)→  101

Mol% C3H8 (7667)→  102

Mol% CH4 (7668)→  102

Mol% Cl2 (7707)→  10297

Description of device parameters

98

Proline Prowirl 200 FOUNDATION Fieldbus

Mol% CO (7669)→  103

Mol% CO2 (7670)→  103

Mol% H2 (7671)→  104

Mol% H2O (7672)→  104

Mol% H2S (7673)→  104

Mol% HCl (7674)→  105

Mol% He (7675)→  105

Mol% i-C4H10 (7676)→  106

Mol% i-C5H12 (7677)→  106

Mol% Kr (7678)→  106

Mol% N2 (7679)→  107

Mol% n-C10H22 (7680)→  107

Mol% n-C4H10 (7681)→  108

Mol% n-C5H12 (7682)→  108

Mol% n-C6H14 (7683)→  108

Mol% n-C7H16 (7684)→  109

Mol% n-C8H18 (7685)→  109

Mol% n-C9H20 (7686)→  109

Mol% Ne (7687)→  110

Mol% NH3 (7688)→  110

Mol% O2 (7689)→  111

Mol% SO2 (7691)→  111

Mol% Xe (7692)→  111

Mol% other gas (7690)→  112

Relative humidity (7731)→  112Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Gas type

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Gas type (7714)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Single gas option is selected.

Description

Use this function to select the type of gas for the measuring application.

Selection• Hydrogen H2• Helium He• Neon Ne• Argon Ar• Krypton Kr• Xenon Xe• Nitrogen N2• Oxygen O2• Chlorine Cl2• Ammonia NH3• Carbon monoxide CO• Carbon dioxide CO2• Sulfur dioxide SO2• Hydrogen sulfide H2S• Hydrogen chloride HCl• Methane CH4• Ethane C2H6
• Propane C3H8
• Butane C4H10• Ethylene C2H4
• Vinyl Chloride C2H3Cl

Factory setting

Methane CH4

Gas mixture

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Gas mixture (7640)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.

Description

Use this function to select the gas mixture for the measuring application.

Selection• Hydrogen H2• Helium He• Neon Ne• Argon Ar• Krypton Kr• Xenon Xe• Nitrogen N2

Endress+Hauser

99Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Oxygen O2• Chlorine Cl2• Ammonia NH3• Carbon monoxide CO• Carbon dioxide CO2• Sulfur dioxide SO2• Hydrogen sulfide H2S• Hydrogen chloride HCl• Methane CH4• Ethane C2H6
• Propane C3H8
• Butane C4H10• Ethylene C2H4
• Vinyl Chloride C2H3Cl• Others
Factory setting

Methane CH4

Mol% Ar

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% Ar (7663)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Argon Ar option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% C2H3Cl

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% C2H3Cl (7664)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Vinyl Chloride C2H3Cl option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %100Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

Description of device parameters

0%

Mol% C2H4

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% C2H4 (7665)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Ethylene C2H4 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% C2H6

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% C2H6 (7666)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Ethane C2H6 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting0%

Endress+Hauser

101Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Mol% C3H8

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% C3H8 (7667)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Propane C3H8 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% CH4



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% CH4 (7668)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Methane CH4 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

100 %Mol% Cl2

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% Cl2 (7707)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Chlorine Cl2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

102

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User entry

0 to 100 %Factory setting0%

Description of device parameters

Mol% CONavigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% CO (7669)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Carbon monoxide CO option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% CO2



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% CO2 (7670)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Carbon dioxide CO2 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting0%

Endress+Hauser

103Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Mol% H2

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% H2 (7671)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Hydrogen H2 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the AGA Nx19 option is not selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% H2O

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% H2O (7672)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% H2S

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% H2S (7673)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Hydrogen sulfide H2S option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

104

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User entry

0 to 100 %Factory setting0%

Description of device parameters

Mol% HCl

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% HCl (7674)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Hydrogen chloride HCl option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% HeNavigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% He (7675)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Helium He option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting0%

Endress+Hauser

105Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Mol% i-C4H10

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% i-C4H10 (7676)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% i-C5H12Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% i-C5H12 (7677)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% Kr



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% Kr (7678)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Krypton Kr option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

106

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Mol% N2

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% N2 (7679)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Nitrogen N2 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the AGA Nx19 option or the ISO 122132 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% n-C10H22



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C10H22 (7680)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting0%

Endress+Hauser

107Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Mol% n-C4H10

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C4H10 (7681)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Butane C4H10 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.
• Or
In the Select medium parameter (→  85), the Liquid option is selected and in the
Select liquid type parameter (→  87), the LPG option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% n-C5H12Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C5H12 (7682)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% n-C6H14

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C6H14 (7683)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

108

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User entry

0 to 100 %Factory setting0%

Description of device parameters

Mol% n-C7H16

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C7H16 (7684)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% n-C8H18

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C8H18 (7685)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% n-C9H20

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% n-C9H20 (7686)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Natural gas option is selected.
• In the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

Endress+Hauser

109

Description of device parameters

User entry

0 to 100 %Factory setting0%

Proline Prowirl 200 FOUNDATION Fieldbus

Mol% NeNavigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% Ne (7687)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Neon Ne option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% NH3



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% NH3 (7688)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Ammonia NH3 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

110

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Mol% O2



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% O2 (7689)

Prerequisite

The following conditions are met:
In the Select medium parameter (→  85), the Gas option is selected.
– In the Select gas type parameter (→  86), the Gas mixture option is selected and in
the Gas mixture parameter (→  99), the Oxygen O2 option is selected.
Or– In the Select gas type parameter (→  86), the Natural gas option is selected and in
the Density calculation parameter (→  88), the ISO 12213- 2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% SO2



Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% SO2 (7691)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Sulfur dioxide SO2 option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting

0%

Mol% Xe

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% Xe (7692)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Xenon Xe option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Endress+Hauser

111Description of device parameters

Factory setting

Proline Prowirl 200 FOUNDATION Fieldbus

0%

Mol% other gas

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Mol% other gas (7690)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Gas mixture option is selected.
• In the Gas mixture parameter (→  99), the Others option is selected.

Description

Use this function to enter the amount of the gas constituent in the gas mixture.

User entry

0 to 100 %Factory setting0%

Relative humidity

Navigation

 Expert → Sensor → Measurement mode → Medium property → Gas composition→ Rel. humidity (7731)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85), the Gas option is selected.
• In the Select gas type parameter (→  86), the Air option is selected.

Description

Use this function to enter the humidity content of the air in %.

User entry

0 to 100 %Factory setting

0%

3.2.5
Navigation"External compensation" submenu
 Expert → Sensor → External comp.

‣ External compensation

112

External value (7622)→  113

Atmospheric pressure (7601)→  113

Delta heat calculation (7736)→  114

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Fixed density (7627)→  114

Fixed temperature (7628)→  114

2nd temperature delta heat (7625)→  115

Fixed process pressure (7629)→  115

Steam quality (7605)→  116

Steam quality value (7630)→  116

External value

Navigation

 Expert → Sensor → External comp. → External value (7622)

Prerequisite

For the following order code:"Sensor version", option "Mass flow"

Description

Use this function to select the process variable which is taken from an external device.
For detailed information on setting the parameter in steam applications, see theSpecial Documentation for the Wet Steam Detection application package.

Selection• Off• Pressure• Relative pressure• Density• Temperature• 2nd temperature delta heat

Factory setting

Off

Atmospheric pressureNavigation

 Expert → Sensor → External comp. → Atmosph. press. (7601)

Prerequisite

In the External value parameter (→  113), the Relative pressure option is selected.

Description

Use this function to enter the value for the ambient pressure to be used for pressure
correction.

User entry

0 to 250 bar

Factory setting

1.01325 bar

Endress+Hauser

113

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Dependency
The unit is taken from the Pressure unit parameter (→  74)

Delta heat calculation



Navigation

 Expert → Sensor → External comp. → Delta heat calc. (7736)

Prerequisite

The Delta heat calculation parameter (→  114) is visible.

Description

Use this function to select the option for calculating the heat transferred via a heatexchanger (=delta heat).

Selection• Off• Device on cold side
• Device on warm side

Factory setting

Device on warm side

Fixed density

Navigation

 Expert → Sensor → External comp. → Fixed density (7627)

Prerequisite

For the following order code:"Sensor version", option "Volume flow"

Description

Use this function to enter a fixed value for the density.

User entry

0.01 to 15 000 kg/m³

Factory setting

1 000 kg/m³

Additional information

Description
The density entered is used to linearize the measured error in the lower Reynolds number
range if the calculated density is not available e.g. "Volume flow" sensor version or the fluid
is a user-specific gas (see table).
Dependency
The unit is taken from the Density unit parameter (→  79)

Fixed temperature

Navigation

 Expert → Sensor → External comp. → Fixed temp. (7628)

Description

Use this function to enter a fixed value for the process temperature.

114

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User entry–200 to 450 °C

Factory setting

20 °C

Additional information

Dependency

Description of device parameters

The unit is taken from the Temperature unit parameter (→  75)

2nd temperature delta heat

Navigation

 Expert → Sensor → External comp. → 2ndTempDeltaHeat (7625)

Prerequisite

The 2nd temperature delta heat parameter (→  115) is visible.

Description

Use this function to enter the second temperature value for calculating the delta heat.

User entry–200 to 450 °C

Factory setting

20 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75)

Fixed process pressureNavigation

 Expert → Sensor → External comp. → Fix. proc.press. (7629)

Prerequisite

The following conditions are met:
• Order code for "Sensor version", option "Mass flow (integrated temperature
measurement)"• In the External value parameter (→  113) the Pressure option is not selected.

Description

Use this function to enter a fixed value for the process pressure.

User entry

0 to 250 bar abs.

Factory setting

0 bar abs.

Additional information

User entry
For detailed information on setting the parameter in steam applications, see theSpecial Documentation for the Wet Steam Detection application package.
Dependency
The unit is taken from the Pressure unit parameter (→  74)

Endress+Hauser

115

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Steam quality



Navigation

 Expert → Sensor → External comp. → Steam quality (7605)

Prerequisite

The following conditions are met:
• Order code for "Application package":– Option ES "Wet steam detection"– Option EU "Wet steam measurement"• In the Select medium parameter (→  85) the Steam option is selected.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Use this function to select the compensation mode for the steam quality.

Selection• Fixed value• Calculated value

Factory setting

Fixed value

Additional information

Selection
For detailed information on setting the parameter in steam applications, see theSpecial Documentation for the Wet Steam Detection application package.

Steam quality value

Navigation

 Expert → Sensor → External comp. → Steam qual. val. (7630)

Prerequisite

The following conditions are met:
• In the Select medium parameter (→  85) the Steam option is selected.
• In the Steam quality parameter (→  116) the Fixed value option is selected.

Description

Use this function to enter a fixed value for the steam quality.

User entry

0 to 100 %Factory setting

100 %Additional information

User entry
For detailed information on setting the parameter in steam applications, see theSpecial Documentation for the Wet Steam Detection application package.

116

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

3.2.6

Description of device parameters"Sensor adjustment" submenu

Navigation

 Expert → Sensor → Sensor adjustm.

‣ Sensor adjustment
Inlet configuration (7641)→  117

Inlet run (7642)→  117

Mating pipe diameter (7648)→  118

Installation factor (7616)→  119

Inlet configuration



Navigation

 Expert → Sensor → Sensor adjustm. → Inlet config. (7641)

Prerequisite

The inlet run correction feature:• Is a standard feature and can only be used in Prowirl F 200.
• Can be used for the following pressure ratings and nominal diameters:
DN 15 to 150 (1 to 6")– EN (DIN)– ASME B16.5, Sch. 40/80

Description

Use this function to select the inlet configuration.

Selection• Off• Single elbow• Double elbow• Double elbow 3D• Reduction

Factory setting

Off

Inlet run

Navigation

 Expert → Sensor → Sensor adjustm. → Inlet run (7642)

Prerequisite

The inlet run correction feature:• Is a standard feature and can only be used in Prowirl F 200.
• Can be used for the following pressure ratings and nominal diameters:
DN 15 to 150 (1 to 6")– EN (DIN)– ASME B16.5, Sch. 40/80

Description

Use this function to enter the length of the straight inlet run.

Endress+Hauser

117Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

User entry

0 to 20 m

Factory setting

0mAdditional information

Dependency
The unit is taken from the Length unit parameter (→  81)

Mating pipe diameter

Navigation

 Expert → Sensor → Sensor adjustm. → D mating pipe (7648)

Description

Use this function to enter the diameter of the mating pipe to enable diameter mismatchcorrection.

User entry

0 to 1 m (0 to 3 ft)

Factory setting

Country-specific:•0m• 0 ftAdditional information

Description
The device has diameter mismatch correction. This can be enabled by entering the actualinternal diameter of the mating pipe in the Mating pipe diameter parameter.
User entry
If the value entered is 0, diameter mismatch correction is disabled. If the standard internal
diameter of the ordered process connection differs from the internal diameter of themating pipe, an additional measuring uncertainty of up to 2 % must be expected if
diameter mismatch correction is disabled.
Limit values
Diameter mismatch correction should be enabled only within the following limit values:
Flange connection:• DN 15 (½"): ±20 % of the internal diameter• DN 25 (1"): ±15 % of the internal diameter• DN 40 (1½"): ±12 % of the internal diameter• DN ≥ 50 (2"): ±10 % of the internal diameter
Disc (wafer version):• DN 15 (½"): ±15 % of the internal diameter• DN 25 (1"): ±12 % of the internal diameter• DN 40 (1½"): ±9 % of the internal diameter• DN ≥ 50 (2"): ±8 % of the internal diameter
Dependency
The unit is taken from the Length unit parameter (→  81)

118

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Installation factor



Navigation

 Expert → Sensor → Sensor adjustm. → Install. factor (7616)

Description

Use this function to enter the factor to adjust installation conditions.

User entry

Positive floating-point number

Factory setting

1.0

Additional information

Description
The calculated volume flow and all measured variables derived from this are multiplied by
the installation factor.

3.2.7"Calibration" submenu

Navigation

 Expert → Sensor → Calibration‣ Calibration
Calibration factor (7604)→  119

Meter body properties (7658)→  120

Calibration factor
Navigation

 Expert → Sensor → Calibration → Cal. factor (7604)

Description

Displays the calibration factor. The calibration factor is determined during device
calibration.

User interface

Positive floating-point number

Factory settingThis value is always > 0 when the device is delivered from the factory.

Additional information

Description
Factor by which the measured vortex frequency must be divided in order to calculate thevolume flow.
Unit
In 1/m³, or vortex pulses per cubic meter

Endress+Hauser

119Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Meter body properties
Navigation

 Expert → Sensor → Calibration → Meter body prop. (7658)

Description

Displays informative text about the measuring tube.

User interface

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /)Factory setting--------------

Additional information

Description
Summarized information about the meter body.
ExampleDN25F-PN40: nominal diameter DN25, flange type, pressure rating 40 bar

3.3
Navigation"Output" submenu
 Expert → Output‣ Output‣ Pulse/frequency/switch output

3.3.1
Navigation→  120"Pulse/frequency/switch output" submenu
 Expert → Output → PFS output‣ Pulse/frequency/switch output

120

Operating mode (0469)→  121

Assign pulse output (0460)→  123

Value per pulse (0455)→  124

Pulse width (0452)→  124

Failure mode (0480)→  125

Pulse output (0456)→  126

Assign frequency output (0478)→  127

Minimum frequency value (0453)→  127

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Maximum frequency value (0454)→  128

Measuring value at minimum
frequency (0476)→  128

Measuring value at maximum
frequency (0475)→  129

Damping output (0477)→  130

Response time (0491)→  130

Failure mode (0451)→  131

Failure frequency (0474)→  132

Output frequency (0471)→  132

Switch output function (0481)→  132

Assign diagnostic behavior (0482)→  133

Assign limit (0483)→  134

Switch-on value (0466)→  135

Switch-off value (0464)→  136

Assign flow direction check (0484)→  136

Assign status (0485)→  137

Switch-on delay (0467)→  137

Switch-off delay (0465)→  137

Failure mode (0486)→  138

Switch status (0461)→  138

Invert output signal (0470)→  138

Operating modeNavigation

 Expert → Output → PFS output → Operating mode (0469)

Description

Use this function to select the operating mode of the output as a pulse, frequency or switch
output.

Endress+Hauser

121

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Pulse• Frequency• Switch

Factory setting

Pulse

Additional information"Pulse" option
Quantity-dependent pulse with configurable pulse width• Whenever a specific volume, corrected volume, mass, total mass, energy or heat isreached (pulse value), a pulse is output, the duration of which was set previously (pulse
width).
• The pulses are never shorter than the set duration.
Example• Total flow approx. 100 g/s• Pulse value 0.1 g• Pulse width 0.05 ms• Pulse rate 1 000 Impuls/s
U[V]

B<P
B

24
t

0
P

A0026883

4
BP

Quantity-proportional pulse (pulse value) with pulse width to be configured
Pulse width entered
Pauses between the individual pulses"Frequency" option
Flow-proportional frequency output with 1:1 on/off ratio
An output frequency is output that is proportional to the value of a process variable, suchas volume flow, corrected volume flow, mass flow, flow velocity, temperature, calculated
saturated steam pressure, steam quality, total mass flow, energy flow or heat flow
difference.
Example• Total flow approx. 100 g/s• Max. frequency 10 kHz• Flow rate at max. frequency1 000 g/s• Output frequency approx. 1 000 Hz
U[V]
24
0

t
A0026886

5

Flow-proportional frequency output"Switch" option
Contact for displaying a condition (e.g. alarm or warning if a limit value is reached)

122

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

ExampleAlarm response without alarm
U[V]
24
0

t
A0026884

6

No alarm, high level

ExampleAlarm response in case of alarm
U[V]
24
0

t
A0026885

7

Alarm, low level

Assign pulse output

Navigation

 Expert → Output → PFS output → Assign pulse (0460)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  121).

Description

Use this function to select the process variable for the pulse output.

Selection• Off• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Energy flow *• Heat flow difference *Factory setting

Volume flow*

Visibility depends on order options or device settings

Endress+Hauser

123Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Value per pulseNavigation

 Expert → Output → PFS output → Value per pulse (0455)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  121), and one of thefollowing options is selected in the Assign pulse output parameter (→  123):• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the value for the measured value that a pulse is equivalent to.

User entry

Positive floating-point number

Factory setting

Depends on country and nominal diameter →  295

Additional information

User entry
Weighting of the pulse output with a quantity.
The lower the pulse value, the• better the resolution.
• the higher the frequency of the pulse response.

Pulse widthNavigation

 Expert → Output → PFS output → Pulse width (0452)

Prerequisite

In the Operating mode parameter (→  121), the Pulse option is selected and one of thefollowing options is selected in the Assign pulse output parameter (→  123):• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the duration of the output pulse.

User entry

5 to 2 000 ms

Factory setting

100 ms*124

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description• Define how long a pulse is (duration).
• The maximum pulse rate is defined by fmax = 1 / (2 × pulse width).
• The interval between two pulses lasts at least as long as the set pulse width.
• The maximum flow is defined by Qmax = fmax × pulse value.
• If the flow exceeds these limit values, the measuring device displays the diagnostic
message S443 Pulse output 1 to 2.
U[V]

U[V]

B<P

B=P
B

B

t
P

t
P
A0026882

B
P

Pulse width entered
Intervals between the individual pulses

Example• Pulse value: 0.1 g• Pulse width: 0.1 ms• fmax: 1 / (2 × 0.1 ms) = 5 kHz• Qmax: 5 kHz × 0.1 g = 0.5 kg/s
The pulse width is not relevant for Automatic pulse option.

Failure modeNavigation

 Expert → Output → PFS output → Failure mode (0480)

Prerequisite

In the Operating mode parameter (→  121), the Pulse option is selected and one of thefollowing options is selected in the Assign pulse output parameter (→  123):• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to select the failure mode of the pulse output in the event of a device
alarm.

Selection• Actual value• No pulses

Factory setting

No pulses*

Visibility depends on order options or device settings

Endress+Hauser

125Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
The dictates of safety render it advisable to ensure that the pulse output shows apredefined behavior in the event of a fault.
Options• Actual value
In the event of a device alarm, the pulse output continues on the basis of the current
flow measurement. The fault is ignored.
• No pulses
In the event of a device alarm, the pulse output is "switched off".
NOTICE! A device alarm is a measuring device error that must be taken seriously. It canaffect the measurement quality such that the quality can no longer be guaranteed. The
Actual value option is only recommended if it can be guaranteed that all possible alarm
conditions will not affect the measurement quality.

Pulse output
Navigation

 Expert → Output → PFS output → Pulse output (0456)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  121).

Description

Displays the pulse frequency currently output.

User interface

Positive floating-point number

Additional information

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.
• The Value per pulse parameter (→  124) and the Pulse width parameter (→  124)
can be used to define the value (i.e. the amount of the measured value that corresponds
to a pulse) and the duration of the pulse.
1
NC
0

1
NO
0
A0025816-EN

0
1
NC
NO

126

Non-conductive
Conductive
NC contact (normally closed)
NO contact (normally open)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

The output behavior can be reversed via the Invert output signal parameter (→  138),
i.e. the transistor does not conduct for the duration of the pulse.
In addition, the behavior of the output in the event of an error (Failure mode parameter(→  125)) can be configured.

Assign frequency output

Navigation

 Expert → Output → PFS output → Assign freq. (0478)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121).

Description

Use this function to select the process variable for the frequency output.

Selection• Off• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Factory setting

Off

Minimum frequency value

Navigation

 Expert → Output → PFS output → Min. freq. value (0453)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the start value frequency.

User entry

0 to 1 000 Hz*

Visibility depends on order options or device settings

Endress+Hauser

127

Description of device parameters

Factory setting

Proline Prowirl 200 FOUNDATION Fieldbus

0 Hz

Maximum frequency value

Navigation

 Expert → Output → PFS output → Max. freq. value (0454)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the end value frequency.

User entry

0 to 1 000 Hz

Factory setting

1 000 Hz

Measuring value at minimum frequency

Navigation

 Expert → Output → PFS output → Val. at min.freq (0476)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the measured value for the start value frequency.

User entry

Signed floating-point number*

128

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

Depends on country and nominal diameter

Additional information

Dependency

Description of device parameters

The entry depends on the process variable selected in the Assign frequency output
parameter (→  127).

Measuring value at maximum frequency

Navigation

 Expert → Output → PFS output → Val. at max.freq (0475)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the measured value for the end value frequency.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

Additional information

Description
Use this function to enter the maximum measured value at the maximum frequency. The
selected process variable is output as a proportional frequency.
Dependency
The entry depends on the process variable selected in the Assign frequency output
parameter (→  127).

*

Visibility depends on order options or device settings

Endress+Hauser

129

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Damping output

Navigation

 Expert → Output → PFS output → Damping out. (0477)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the reaction time of the output signal to fluctuations in themeasured value.

User entry

0 to 999.9 s

Factory setting

5.0 s

Additional information

Description
Use this function to enter a time constant (PT1 element) for frequency output damping.
The frequency output is subject to separate damping that is independent of all precedingtime constants.

Response time
Navigation

 Expert → Output → PFS output → Response time (0491)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Displays the response time. This specifies how quickly the pulse/frequency/switch output
reaches the measured value change of 63 % of 100 % of the measured value change.

*

130

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User interface

Positive floating-point number

Additional information

Description

Description of device parameters

The response time is made up of the time specified for the following dampings:• Damping of pulse/frequency/switch output
and• Depending on the measured variable assigned to the output.
Flow damping

Failure modeNavigation

 Expert → Output → PFS output → Failure mode (0451)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to select the failure mode of the frequency output in the event of a device
alarm.

Selection• Actual value• Defined value• 0 HzFactory setting

0 Hz

Additional information

Selection• Actual value
In the event of a device alarm, the frequency output continues on the basis of the current
flow measurement. The fault is ignored.
• Defined value
In the event of a device alarm, the frequency output continues on the basis of apredefined value. This Failure frequency (→  132) replaces the current measured
value and the alarm can be bypassed in this way. The actual measurement is switched
off for the duration of the alarm.
• 0 HzIn the event of a device alarm, the frequency output is "switched off".
NOTICE! A device alarm is a measuring device error that must be taken seriously. It canaffect the measurement quality such that the quality can no longer be guaranteed. The
Actual value option is only recommended if it can be guaranteed that all possible alarm
conditions will not affect the measurement quality.

*

Visibility depends on order options or device settings

Endress+Hauser

131Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Failure frequency

Navigation

 Expert → Output → PFS output → Failure freq. (0474)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121), and oneof the following options is selected in the Assign frequency output parameter(→  127):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter the value for the frequency output in the event of a device alarmin order to bypass the alarm.

User entry

0.0 to 1 250.0 Hz

Factory setting

0.0 HzOutput frequency
Navigation

 Expert → Output → PFS output → Output freq. (0471)

Prerequisite

In the Operating mode parameter (→  121), the Frequency option is selected.

Description

Displays the actual value of the output frequency which is currently measured.

User interface

0 to 1 250 Hz

Switch output function



Navigation

 Expert → Output → PFS output → Switch out funct (0481)

Prerequisite

In the Operating mode parameter (→  121) the Switch option is selected.

Description

Use this function to select a function for the switch output.

*

132

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

• Off
• On
• Diagnostic behavior• Limit• Status

Factory setting

Off

Additional information

Selection

Description of device parameters• Off
The switch output is permanently switched off (open, non-conductive).
• On
The switch output is permanently switched on (closed, conductive).
• Diagnostic behavior
Indicates if the diagnostic event is present or not. Is used to output diagnostic
information and to react to it appropriately at the system level.
• Limit
Indicates if a specified limit value has been reached for the process variable. Is used to
output diagnostic information relating to the process and to react to it appropriately at
the system level.
• Status
Indicates the device status depending on whether empty pipe detection or low flow cut
off is selected.

Assign diagnostic behavior

Navigation

 Expert → Output → PFS output → Assign diag. beh (0482)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Diagnostic behavior option is selected in the Switch output function parameter(→  132).

Description

Use this function to select the diagnostic event category that is displayed for the switch
output.

Selection• Alarm• Alarm or warning• Warning

Factory setting

Alarm

Additional information

Description
If no diagnostic event is pending, the switch output is closed and conductive.
Options• Alarm
The switch output signals only diagnostic events in the alarm category.
• Alarm or warning
The switch output signals diagnostic events in the alarm and warning category.
• Warning
The switch output signals only diagnostic events in the warning category.

Endress+Hauser

133

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign limitNavigation

 Expert → Output → PFS output → Assign limit (0483)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Limit option is selected in the Switch output function parameter (→  132).

Description

Use this function to select a process variable for the limit function.

Selection• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Energy flow *• Heat flow difference *• Reynolds number *• Totalizer 1• Totalizer 2• Totalizer 3

Factory setting

Volume flow

Additional information

Description
Behavior of status output when Switch-on value > Switch-off value:• Process variable > Switch-on value: transistor is conductive• Process variable < Switch-off value: transistor is non-conductive
A
1
2
t
B
3
4

t
A0026891

1
2
3
4
A
B

Switch-on value
Switch-off value
Conductive
Non-conductive
Process variable
Status output

Behavior of status output when Switch-on value < Switch-off value:• Process variable < Switch-on value: transistor is conductive• Process variable > Switch-off value: transistor is non-conductive*

134

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

A
1
2
t
B
3
4

t
A0026892

1
2
3
4
A
B

Switch-off value
Switch-on value
Conductive
Non-conductive
Process variable
Status output

Behavior of status output when Switch-on value = Switch-off value:• Process variable > Switch-on value: transistor is conductive• Process variable < Switch-off value: transistor is non-conductive
A

1
t
B
2
3

t
A0026893

1
2
3
A
B

Switch-on value = Switch-off value
Conductive
Non-conductive
Process variable
Status output

Switch-on value

Navigation

 Expert → Output → PFS output → Switch-on value (0466)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Limit option is selected in the Switch output function parameter (→  132).

Description

Use this function to enter the measured value for the switch-on point.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 m³/h• 0 ft³/hEndress+Hauser

135

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
Use this function to enter the limit value for the switch-on value (process variable >switch-on value = closed, conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit depends on the process variable selected in the Assign limit parameter(→  134).

Switch-off value

Navigation

 Expert → Output → PFS output → Switch-off value (0464)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Limit option is selected in the Switch output function parameter (→  132).

Description

Use this function to enter the measured value for the switch-off point.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 m³/h• 0 ft³/h

Additional information

Description
Use this function to enter the limit value for the switch-off value (process variable <switch-off value = open, non-conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit depends on the process variable selected in the Assign limit parameter(→  134).

Assign flow direction checkNavigation

 Expert → Output → PFS output → Assign dir.check (0484)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Flow direction check option is selected in the Switch output function parameter(→  132).

Description

Use this function to select a process variable for monitoring the flow direction.

Selection• Off• Volume flow
• Mass flow
• Corrected volume flow

136

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

Description of device parameters

Volume flow

Assign status



Navigation

 Expert → Output → PFS output → Assign status (0485)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Status option is selected in the Switch output function parameter (→  132).

Description

Use this function to select a device status for the switch output.

Selection• Low flow cut off• Digital output 6

Factory setting

Low flow cut off

Additional information

Options
If empty pipe detection or low flow cut off are enabled, the output is conductive.
Otherwise, the switch output is non-conductive.

Switch-on delay

Navigation

 Expert → Output → PFS output → Switch-on delay (0467)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Limit option is selected in the Switch output function parameter (→  132).

Description

Use this function to enter a delay time for switching on the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Switch-off delay

Navigation

 Expert → Output → PFS output → Switch-off delay (0465)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  121).
• The Limit option is selected in the Switch output function parameter (→  132).

Description

Use this function to enter a delay time for switching off the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Endress+Hauser

137

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Failure modeNavigation

 Expert → Output → PFS output → Failure mode (0486)

Description

Use this function to select a failsafe mode for the switch output in the event of a device
alarm.

Selection• Actual status• Open• Closed

Factory setting

Open

Additional information

Options• Actual status
In the event of a device alarm, faults are ignored and the current behavior of the input
value is output by the switch output. The Actual status option behaves in the same way
as the current input value.
• OpenIn the event of a device alarm, the switch output's transistor is set to non-conductive.
• ClosedIn the event of a device alarm, the switch output's transistor is set to conductive.

Switch status
Navigation

 Expert → Output → PFS output → Switch status (0461)

Prerequisite

The Switch option is selected in the Operating mode parameter (→  121).

Description

Displays the current switch status of the status output.

User interface• Open• Closed

Additional information

User interface• OpenThe switch output is not conductive.
• Closed
The switch output is conductive.

Invert output signal

Navigation

 Expert → Output → PFS output → Invert outp.sig. (0470)

Description

Use this function to select whether to invert the output signal.

Selection

• No
• Yes

138

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

No

Additional information

Selection

Description of device parameters

No option (passive - negative)
1
0
A0026693

Yes option (passive - positive)
1
0
A0026692

3.4"Communication" submenu

Navigation

Expert → Communication‣ Communication
Device address (11061)→  139‣ Resource block→  140

Device address
NavigationDescription

Displays the device address.

User interface

1 to 255

Endress+Hauser

Expert → Communication → Device address (11061)

139Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

3.4.1"Resource block" submenu

Navigation

 Expert → Communication → Resource block‣ Resource block
Block tag (10702)→  140

Target mode (10728)→  141

Actual mode (10725)→  143

Manufacturer Id (10721)→  146

Device type (10711)→  146

Device Revision (10710)→  147

DD Revision (10709)→  147

Restart (10800)→  148

Write Lock (10747)→  154

ITK Version (10794)→  161

Block tag
Navigation

 Expert → Communication → Resource block → Block tag (10702)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Static Revision
NavigationDescription

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

140

Expert → Communication → Resource block → Static Revision (10735)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Communication → Resource block → Tag Description (10736)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Communication → Resource block → Strategy (10734)

Description

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Alert Key
Navigation

Expert → Communication → Resource block → Alert Key (10696)

Description

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
Navigation

 Expert → Communication → Resource block → Target mode (10728)

Description

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Endress+Hauser

141

Description of device parameters

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options

Proline Prowirl 200 FOUNDATION Fieldbus• ROutIn the ROut operating mode, the set point (SP) for the function block is controlled by thefield bus host system, which runs over an interface, via the RIn parameter. The set point(SP) is used as an output signal after internal calculations. The output value and thestatus of the function block is communicated to the field bus host system as feedback via
the ROut parameter. The set point can be initiated or retained for the value of theprocess variables.
• RCasIn the RCas operating mode (external cascade mode), the set point (SP) for the function
block is controlled by the field bus host system, which runs over an interface, via theRCas-In parameter. The set point (SP) is used as an output signal after internal
calculations. The output value and status of the function block is communicated to thefield bus host system as feedback via the RCas-Out parameter.
• Cas
In the Cas (cascade mode) operating mode, the function block receives a discrete signal
via the function block input, which is provided by the Cas-In parameter of an upstream
function block. This signal controls the set point (SP) of the function block and is used as
an output signal after internal calculations. The output is used to communicate theoutput value and status to the upstream function block.
• AutoThe Auto operating mode is the normal operating mode of the function block. The set
point (SP) is specified locally in the function block and is used as an output signal after
internal calculations. This set point can be specified by the user via an interface.
• Man
In the Man operating mode, the output value can be specified directly in the function
block. This is specified by the user via an interface. There is no internal calculation. The
algorithm is initiated in such a way that there is no interruption when the operating
mode is changed. The set point can be retained or initiated for the value of the process
variables or for the set point of the previous operating mode if the system is switchedback to it.

142

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• LO
The LO operating mode is used in control and output blocks that support a track input
parameter. The manufacturer can also provide a local lock switch on the device to enable
the LO operating mode. Tracking must be supported in the group of control parameters
and is initiated by a discrete track-in parameter.
In local bridge mode, the output value of the function block is set in order to track thevalue of the track-input parameter. The algorithm is initiated in such a way that there isno interruption when the operating mode switches from LO back to the previous
operating mode. The set point can be initiated or retained for the value of the process
variables.
• IManIn the IMan operating mode, the output value of the function block is performed as areaction to the status of the back-calculation-input parameter. When this statusindicates that there is no signal for the final output element, the control blocks ensurethat there is a smooth transition. The back-calculation-input parameter is supported by
all control and output blocks. The set point can be initiated or retained for the value ofthe process variables.
• OOS
In the OOS operating mode, implementation of the function block is blocked. Either thelast valid value or, in the case of an output block, the last valid set point is retained as theoutput value. This operating mode is used during configuration of the device.

Actual mode
Navigation

 Expert → Communication → Resource block → Actual mode (10725)

Description

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modethat the function block is currently operating in. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  141).

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

User interface
Detailed description of the options displayed: Target mode parameter (→  141)

Permitted mode
NavigationDescription

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  141) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Endress+Hauser

Expert → Communication → Resource block → Permitted mode (10727)

143Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Normal mode
Navigation

Expert → Communication → Resource block → Normal mode (10726)

Description

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Block Error
NavigationDescription

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance

144

Expert → Communication → Resource block → Block Error (10703)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Resource State
Navigation

Expert → Communication → Resource block → Resource State (10730)

Description

Displays the Resource State: Displays the current operating mode of the resource block.

User interface• Uninitialized• StartRestart• Initialization• Online Linking• Online• Standby• Failure

Additional information

User interface• Uninitialized
The resource block is in an invalid operating mode.
• StartRestart
The resource block is in the start-up or restore phase. The necessary memory andhardware tests are carried out for the current operation. If these tests are successful, theresource block switches to the Initialization operating mode. If these tests are notsuccessful, the resource block switches to the Failure operating mode.
• Initialization
The resource block is in this operating mode if the block successfully passes the
hardware tests from the StartRestart or Failure status. In this operating mode, all alarmsof the function block that are not displayed are automatically confirmed andacknowledged. When system management is operational, the implementation of theblock can be planned and the resource block switches to the Online Linking operatingmode.
• Online Linking
The resource block is in this operating mode when the block switches from theInitialization or Online operating mode. The configured connections between thefunction blocks are not yet established. When all configured connections have been
established, the resource block switches to the Online operating mode.

Endress+Hauser

145

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Online
Normal operating mode, the resource block is in the Auto operating mode. The
configured connections between the function blocks have been established. If one of theconnections could not be established, the resource block switches back to the Online
Linking operating mode.
• Standby
The Resource Block is in the OOS option operating mode. It is not possible to implement
the remaining blocks. The operating mode of the transducer block is not necessarily
affected by this. Switching the resource block to the Auto mode causes the resource
block to switch back to the StartRestart operating mode.
• Failure
The resource block is in the fault state. It switches to this operating mode if a memory or
hardware fault that would hinder current operation has been detected in the block. The
fault can affect the block or the entire device. When this operating mode is active, blocks
with an output function are also in the fault state. The hardware test is carried out again.
If the error does not reoccur, the resource block switches back to the Initialization
operating mode.

DD Resource
Navigation

Expert → Communication → Resource block → DD Resource (10708)

Description

Displays the DD Resource: Indicates the reference source for the device description (DD).

User interface

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Additional information

User interface
Zero: No device description is stored on the device.

Manufacturer Id
Navigation

 Expert → Communication → Resource block → Manufacturer Id (10721)

Description

Manufacturer Id display: is used by the interface to assign the right DD file for the
resource.

User interface

Endress+Hauser

Device type
Navigation

 Expert → Communication → Resource block → Device type (10711)

Description

Use this function to view the device type with which the measuring device is registeredwith FOUNDATION Fieldbus.

User interface

Prowirl 200

146

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Device Revision
Navigation

 Expert → Communication → Resource block → Device Revision (10710)

Description

Use this function to view the device revision with which the measuring device is registeredwith FOUNDATION Fieldbus.

User interface

1

DD Revision
Navigation

 Expert → Communication → Resource block → DD Revision (10709)

Description

Displays the revision number of the device description (DD).

User interface

1

Additional information

Description
This display can be used to ensure that the right system files (DD = device description)
are used for integration into the host system. The system files can be downloaded freeof charge online at: www.endress.com.

Grant
Navigation

Expert → Communication → Resource block → Grant (10718)

Description

Option for releasing certain access authorizations of the field bus host system on thedevice.

Selection• Program• Tune• Alarm• Local• Operate• Service• Diagnostic

Deny
NavigationDescription

Option for restricting certain access authorizations of the field bus host system on thedevice.

Endress+Hauser

Expert → Communication → Resource block → Deny (10717)

147

Description of device parameters

Selection

Proline Prowirl 200 FOUNDATION Fieldbus• Program Denied• Tune Denied• Alarm Denied• Local• Operate Denied

Hard Types
Navigation

Expert → Communication → Resource block → Hard Types (10719)

Description

Displays the input signal type for the Analog input block.

User interface• Scalar Input• Scalar Output• Discrete Input• Discrete Output

RestartNavigation

 Expert → Communication → Resource block → Restart (10800)

Description

Use this function to select a manual restart or a manual device reset.

Selection• Uninitialized• Run• Resource• Defaults• Processor• To factory defaults• To delivery settings• ENP restart• To transducer defaults• Factory Default Blocks

Factory setting

Uninitialized

Features
NavigationDescription

Displays the additional options that are supported by the measuring device.

User interface• Reports• Faultstate• Soft W Lock
• Hard W Lock
• Change Bypass in Auto148

Expert → Communication → Resource block → Features (10713)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• MVC Report Distribution supported• Multi-bit Alarm (Bit-Alarm) Support• Defer Inter-Parameter Write Checks

Feature Selection
Navigation

Expert → Communication → Resource block → Feature Select (10714)

Description

Use this function to select additional options that are supported by the measuring device.

Selection• Reports• Faultstate• Soft W Lock
• Hard W Lock
• Change Bypass in Auto
• MVC Report Distribution supported• Multi-bit Alarm (Bit-Alarm) Support• Defer Inter-Parameter Write Checks

Cycle Type
Navigation

Expert → Communication → Resource block → Cycle Type (10707)

Description

Displays the implementation methods for the block that are supported by the measuring
device.

User interface• Scheduled• Block Execution

Cycle Selection
NavigationDescription

Use this function to select the implementation method for the block that is used by thefield bus host system. This implementation method is selected by the field bus hostsystem.

Selection• Scheduled• Block Execution

Endress+Hauser

Expert → Communication → Resource block → Cycle Selection (10706)

149Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Minumum Cycle Time
Navigation

Expert → Communication → Resource block → Minumum Cyc.Time (10724)

Description

Displays the implementation time of all function blocks that are available in the measuring
device.

User interface

Positive integer

Memory Size
Navigation

Expert → Communication → Resource block → Memory Size (10723)

Description

Displays the available configuration memory in kilobytes.

User interface

0 to 65 535 Kbytes

Nonvolatile Cycle Time
Navigation

Expert → Communication → Resource block → Nonvolat CycTime (10729)

Description

Displays the time interval during which the dynamic device parameters are stored in the
non-volatile memory.

User interface

Positive integer

Free Space
Navigation

Expert → Communication → Resource block → Free Space (10715)

Description

Displays the free system memory volume available for implementing further function
blocks as a percentage.

User interface

0.000000 to 100.000 %Free Time
NavigationDescription

Displays the free system time available for implementing further function blocks as apercentage.

150

Expert → Communication → Resource block → Free Time (10716)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User interface

Description of device parameters0.000000 to 100.000 %Clear Fault State
Navigation

Expert → Communication → Resource block → Clear Fault Stat (10704)

Description

Use this function to select the fault state for the Discrete outputs (→  247) block.

Selection• Uninitialized• Off• Clear

Factory setting

Uninitialized

Additional information

Options
The fault state can be disabled with Clear option.

Confirm Time
Navigation

Expert → Communication → Resource block → Confirm Time (10705)

Description

Use this function to enter a time interval for confirming the event report. If the measuring
device does not receive a confirmation within this interval, the event report is sent to thefield bus host system again.

User entry

Positive integer

Factory setting

0 1/32 ms

Fault State
Navigation

Expert → Communication → Resource block → Fault State (10712)

Description

Displays the current status of the fault state for the Discrete outputs (→  247) block.

User interface• Uninitialized• Clear• Active

Additional information

User interface• Uninitialized• Clear
The fault state is disabled.
• Active
The fault state is enabled.

Endress+Hauser

151

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Limit Notify
Navigation

Expert → Communication → Resource block → Limit Notify (10720)

Description

Use this function to enter the maximum number of event reports that may be pending at
the same time without confirmation.

User entry

0 to 255

Factory setting0

Max Notify
Navigation

Expert → Communication → Resource block → Max Notify (10722)

Description

Displays the maximum number of event reports that are supported by the measuring
device and may be pending at the same time without confirmation.

User interface

0 to 255

Set Fault State
Navigation

Expert → Communication → Resource block → Set Fault State (10731)

Description

Option for manually enabling or disabling the fault state for the Discrete outputs(→  247) function block.

Selection

• OFF
• SET

Factory setting

OFF

Additional information

Options• OFF
The fault state is disabled.
• SET
The fault state is enabled.

Shed Remote Cascade
NavigationDescription

Use this function to enter the time interval after which the computer stops writing tofunction blocks with RCas points.

152

Expert → Communication → Resource block → Shed Remote Casc (10732)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User entry

Positive integer

Factory setting

0 1/32 ms

Description of device parameters

Shed Remote Out
Navigation

Expert → Communication → Resource block → Shed Remote Out (10733)

Description

Use this function to enter the time interval after which the computer stops writing tofunction blocks with ROut points.

User entry

Positive integer

Factory setting

0 1/32 ms

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10740)

Description

Option for manually acknowledging an update of the static block parameter.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new diagnostic event occurs, the measuring device sets Unacknowledged
option.
• If the diagnostic event has been acknowledged, the user can set Acknowledged
option.

Update State
NavigationDescription

Displays the status of an update of the static block parameter. The status indicates
whether the update was communicated or not.

User interface• Uninitialized• Reported• Not Reported

Endress+Hauser

Expert → Communication → Resource block → Update State (10741)

153Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Time Stamp
Navigation

Expert → Communication → Resource block → Time Stamp (10739)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of an update to the static block parameter that has not yet beencommunicated was identified. The time stamp is retained until the update confirmation isreceived.

User interface

Days (d), hours (h), minutes (m), seconds (s)Static revision
Navigation

Expert → Communication → Resource block → Static revision (10738)

Description

Displays the Static revision: Each instance of a static block parameter being accessed withwrite and report access due to an update is counted (event counter).

User interface

0 to 65 535

Relative Index
Navigation

Expert → Communication → Resource block → Relative Index (10737)

Description

Displays the relative index of the static block parameter that triggered the alarm (OD index
minus field bus start index).

User interface

0 to 65 535

Write Lock
Navigation

 Expert → Communication → Resource block → Write Lock (10747)

Description

Option for enabling and disabling the hardware write protection.

Selection• Not Locked• Locked

Factory setting

Not Locked

154

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
Option for enabling and disabling write access to the measuring device via the
FOUNDATION Fieldbus (acyclic data transmission, e.g. via the "FieldCare" operating
program).
For detailed information on hardware write protection, see the "Write protection via
write protection switch" section of the Operating Instructions.
Options• Not Locked
Device data can be changed via the FOUNDATION Fieldbus interface.
• Locked
Device data cannot be changed via the FOUNDATION Fieldbus interface.

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10700)

Description

Option for manually acknowledging a block alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
NavigationDescription

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Endress+Hauser

Expert → Communication → Resource block → Alarm State (10697)

155

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Time Stamp
Navigation

Expert → Communication → Resource block → Time Stamp (10699)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of the block alarm that has not yet been communicated to the field bus hostsystem was detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigation

Expert → Communication → Resource block → Subcode (10698)

Description

Displays the specific cause of the block alarm.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Value
NavigationDescription

Displays the value of the affected parameter at the time at which the block alarm was
detected.

User interface

0 to 255

156

Expert → Communication → Resource block → Value (10701)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Current
Navigation

Expert → Communication → Resource block → Current (10692)

Description

Use this function to view the current status of the process alarms.

User interface• Discrete Alarm• Block Alarm• Fail Alarm• Off Spec Alarm• Maintenance Alarm• Check Alarm

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10694)

Description

Displays an unacknowledged process alarm.

User interface• Disc Alm Unack• Block Alm Unack• Fail Alm Unack• Off Spec Alm Unack• Maint Alm Unack• Check Alm Unack

Unreported
Navigation

Expert → Communication → Resource block → Unreported (10695)

Description

Displays a process alarm that has not been communicated.

User interface• Disc Alm Unrep• Block Alm Unrep• Fail Alm Unrep• Off Spec Alm Unrep• Maint Alm Unrep• Check Alm Unrep

Disabled
NavigationDescription

Option for disabling a process alarm category.

Endress+Hauser

Expert → Communication → Resource block → Disabled (10693)

157

Description of device parameters

Selection

Proline Prowirl 200 FOUNDATION Fieldbus• Disc Alm Disabled• Block Alm Disabled• Fail Alm Disabled• Off Spec Alm Disabled• Maint Alm Disabled• Check Alm Disabled

Acknowledge Option
Navigation

Expert → Communication → Resource block → Ack. Option (10691)

Description

Option for automatic acknowledgment of process alarms in a specific category.

Selection• Disc Alm Auto Ack• Blk Alm Auto Ack• Fail Alm Auto Ack• Off Spec Alm Auto Ack• Maint Alm Auto Ack• Check Alm Auto Ack

Additional information

Description
Use this function to determine whether an alarm must be acknowledged via the field bus
host system.
If the process alarm option has not been enabled in this parameter, this process alarm
must only be acknowledged in Unacknowledged parameter (→  155).
Current parameter (→  157) indicates the current status of all process alarms.

Write Priority
Navigation

Expert → Communication → Resource block → Write Priority (10748)

Description

Use this function to enter the priority for the write protection alarm.

User entry

0 to 15

Factory setting0

Additional information

Description
If write protection is disabled on the FOUNDATION Fieldbus I/O board, the alarm priority
entered here is checked before the status change is transferred to the field bus hostsystem. The alarm priority determines the behavior in the event of an active write
protection alarm.
The alarm is triggered when the write protection is disabled.

158

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10745)

Description

Option for manually acknowledging a write protection alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Communication → Resource block → Alarm State (10742)

Description

Displays the status of the write protection alarm. The status indicates whether the write
protection alarm is active and whether it has already been communicated to the field bus
host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Time Stamp
NavigationDescription

Displays the time stamp indicating when the analysis of the block was started and when
the status change of a write protection alarm that has not yet been communicated was
identified. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Expert → Communication → Resource block → Time Stamp (10744)

159

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

SubcodeNavigation

Expert → Communication → Resource block → Subcode (10743)

Description

Displays the specific cause of the write protection alarm, which is to be communicated to
the field bus host system.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Discrete Value
NavigationDescription

Displays the discrete value of the affected parameter at the time at which the write
protection alarm was detected.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

160

Expert → Communication → Resource block → Discrete Value (10746)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

ITK Version
Navigation

 Expert → Communication → Resource block → ITK Version (10794)

Description

Displays the revision status of the Interoperability Test Kits (ITK).

User interface

6

3.5"Analog inputs" submenu

In the analog input function block (AI function block), the process variables from thetransducer block are prepared in terms of control for the subsequent automation functions(e.g. scaling, limit value processing). The automation function is defined byinterconnecting the outputs.
Navigation

 Expert → Analog inputs‣ Analog inputs‣ Analog input 1 to 4

3.5.1
Navigation→  161"Analog input 1 to 4" submenu
 Expert → Analog inputs → Analog input 1 to 4

‣ Analog input 1 to 4

Endress+Hauser

Block tag (6901–1 to 4)→  162

Status (6906–1 to 4)→  168

Value (6907–1 to 4)→  170

Channel (6902–1 to 4)→  190

Linearization Type (6905–1 to 4)→  191

Process Value Filter Time
(6909–1 to 4)→  196161

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Block tag
Navigation

 Expert → Analog inputs → Analog input 1 to 4 → Block tag (6901–1 to 4)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Static Revision
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Static Revision (6973–1 to 4)

Description

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Tag Description (6974–1 to 4)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
NavigationDescription

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0162

Expert → Analog inputs → Analog input 1 to 4 → Strategy (6972–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Alert Key
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Alert Key (6916–1 to 4)

Description

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Target mode (6960–1 to 4)

Description

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Actual mode
NavigationDescription

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modein which the function block is currently operating. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  163).

User interface• ROut• RCas• Cas• Auto• Man

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Actual mode (6957–1 to 4)163

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• LO• IMan• OOS
Additional information

User interface
Detailed description of the options displayed: Target mode parameter (→  141)

Permitted mode
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Permitted mode (6959–1 to 4)

Description

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  163) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Normal mode
NavigationDescription

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

Auto

164

Expert → Analog inputs → Analog input 1 to 4 → Normal mode (6958–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Block Error
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Block Error (6922–1 to 4)

Description

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

StatusNavigationDescription

Displays the status of the input value (PV).

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Status (6964–1 to 4)

165

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit
166

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState Constant

Value
NavigationDescription

Displays the input value (PV).

User interface

Signed floating-point number

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Value (6965–1 to 4)167

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Dependency
The unit is taken from the: Units index parameter (→  180)

StatusNavigation

 Expert → Analog inputs → Analog input 1 to 4 → Status (6906–1 to 4)

Description

Displays the status of the output value (OUT).

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited

168

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited ConstantEndress+Hauser

169

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState Constant

Value
Navigation

 Expert → Analog inputs → Analog input 1 to 4 → Value (6907–1 to 4)

Description

Input for the output value (OUT).

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Simulate Status
NavigationDescription

Use this function to select the status that is used for the transducer block simulation.

Selection• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited

170

Expert → Analog inputs → Analog input 1 to 4 → Simulate Status (6967–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit
Endress+Hauser

171

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState Constant
Factory setting

172

Bad NonSpecific NotLimited

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Simulate Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Simulate Value (6968–1 to 4)

Description

Use this function to enter the value that is used for the transducer block simulation.

User entry

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  176)

Transducer Status
NavigationDescription

Displays the current status of the transducer block.

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Transducer Stat (6969–1 to 4)

173

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant
174

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState ConstantTransducer Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Transducer Value (6970–1 to 4)

Description

Displays the current value of the transducer block.

User interface

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  176)

Endress+Hauser

175

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Simulate En/Disable
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Sim. En/Disable (6966–1 to 4)

Description

Use this function to enable or disable the function block simulation.

Selection• Uninitialized• Disabled• Active

Factory setting

Uninitialized

EU at 100%
Navigation

Expert → Analog inputs → Analog input 1 to 4 → EU at 100% (6982–1 to 4)

Description

Use this function to enter the upper limit value of the output value measuring range.

User entry

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  176)

EU at 0%Navigation

Expert → Analog inputs → Analog input 1 to 4 → EU at 0% (6981–1 to 4)

Description

Use this function to enter the lower limit value of the output value measuring range.

User entry

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  176)

Units index
NavigationDescription

Use this function to select the unit for the output value.

176

Expert → Analog inputs → Analog input 1 to 4 → Units index (6983–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

Endress+Hauser

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d
• l/s
• l/min• l/d• l/h• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/d• m³/s• m³/min• m³/h• m³/d• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• ml/s• ml/min• ml/h• ml/d• Nl/s• Nl/d• Nl/h• Nl/min• Nm³/s• Nm³/min• Nm³/h• Nm³/d• Sm³/s• Sm³/min• Sm³/h• Sm³/d• MSft³/d• kg/l• kg/m³• kg/dm³• g/cm³• g/m³

Description of device parameters

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)• Sft³/s• Sft³/min• Sft³/h• Sft³/d• MSft³/D• lb/ft³

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/d (imp;oil)• bbl/h (imp;oil)• bbl/min (imp;oil)• bbl/s (imp;oil)• lb/gal (imp)• lb/bbl (imp;oil)• MMBtu/min• MMBtu/h• MMBtu/d• MBtu/s• MBtu/min• MBtu/h• MBtu/d• MMBtu/s• Btu/s• Btu/min• Btu/h• Btu/day• Mgal (imp)• Sgal (imp)• bbl (imp;oil)• gal (imp)• MBtu• Btu• MMBtu177

Description of device parameters• SD4°C• SD15°C• SD20°C• SG4°C• SG20°C• SG15°C
•K
• °C
• m/s
• bar
• mbar a
• Pa
• MPa
• kPa
• torr
• atm
•%
• kW
• MW
• MJ/h• GJ/d• Gcal/h• MJ/s• MJ/min• MJ/d• kJ/s• kJ/min• kJ/h• kJ/d• GJ/s• GJ/min• GJ/h• kcal/s• kcal/min• kcal/h• kcal/d• Mcal/s• Mcal/min• Mcal/h• Mcal/d• Gcal/s• Gcal/min• Gcal/d
•W
• TW
• GW
• mW
• µW
• nW
• pW
•g
• Nl
• Nm³
• Sl
• Sm³
• ml
• Ml Mega
• kg
•l
178

Proline Prowirl 200 FOUNDATION Fieldbus• lb/gal (us)• lb/bbl (us;tank)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)
• °F
• °R
• ft/s• psi• inH2Og(4°C)• inH2Og(68°F)• ftH2Og(68°F)
• hp
• oz
• af
• fl oz (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;tank)• Sgal (us)• Sbbl (us;liq.)• Sft³• kgal (us)
• lb
• ft³
• gal (us)• bbl (us;beer)• STonEndress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

•t
• m³
• dm³
• cm³
• hl
• kWh
• MWh
• MJ
• GJ
• kgf/cm²• GWh• kcal• Mcal• gf/cm²• kJ• GcalOther units• mmH2O (4°C)• mmH2O (68°F)• inHg (0°C)• mmHg (0°C)• mmH2Oa(4°C)• mmH2Oa(68°F)• inH2O (4°C)• inH2O (68°F)• ftH2O (4°C)• ftH2O (68°F)
Factory setting%

Additional information

ResultThe selected unit applies for:• Simulate Value parameter (→  173)
• Transducer Value parameter (→  175)
• EU at 0% parameter (→  176)
• EU at 100% parameter (→  176)

DecimalNavigationDescription

Use this function to enter the number of decimal places for the output value.

User entry–128 to 127

Factory setting0

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Decimal (6980–1 to 4)

179

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

EU at 100%
Navigation

Expert → Analog inputs → Analog input 1 to 4 → EU at 100% (6963–1 to 4)

Description

Use this function to enter the upper limit value of the input value measuring range fromthe transducer block (input value).

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

EU at 0%Navigation

Expert → Analog inputs → Analog input 1 to 4 → EU at 0% (6962–1 to 4)

Description

Function for entering the lower limit value of the input value measuring range from thetransducer block (input value).

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Units index
NavigationDescription

Use this function to select the unit for the input value from the transducer block (input
value).

180

Expert → Analog inputs → Analog input 1 to 4 → Units index (6908–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection

Endress+Hauser

SI units•K•m• m³• Pa s• m²/s
•P
• cP
• St
• cSt
• Wbm
• Ns/m
• 1/Jm³• e/Vm³• m³/C• (1/32 millisec)/min
• °C
•'
•'
• rad
•°
• gon
• µm
• mm
• cm³
• dm³
• hl
•l
• ml
•s
• min
•d
•h
• ks
• µs
• ms
• m/h
• m/s
• mm/s
• Hz
•g
• kg
• GWh
•J
• kWh
• MWh
• kcal• Mcal• kW• MW•W• MJ/h
• mV
• Ohm
• pF
•V
• Ml Mega• mbar a• bar

Description of device parameters

US units• mils
• °F
• °R
• ft
• in
• bbl (us;liq.)• Sft³• in/min• oz• STon• lb/d• lb/h• lb/min• lb/s• STon/d• STon/h• STon/min• STon/s• mile
• yd
• ft²
• in²
• mile²• yd²• ft³• gal (us)• quart• pint• yd³• mile³• in³• bushel• bbl (us;oil)• ft/s• ft/h• lb• ft/min• in/h• in/s• yd/h• yd/min• yd/s• lb/in³• lb/gal (us)
• STon/yd³• psi• psi a• psi g• ftlbf• hp• lb/ft³• MPH• ft³/d• ft³/h• ft³/min• ft³/s• Sft³/h• Sft³/min

Imperial units• lbf/in• gal (imp)• Btu• LTon• datherm• Btu/h• Btu/day• Btu/min• Btu/s• LTon/d• LTon/h• LTon/min• LTon/s• gal/d (imp)• gal/h (imp)• gal/min (imp)• gal/s (imp)• lb/gal (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• kgal/d (imp)• kgal/h (imp)• kgal/min (imp)• kgal/s (imp)• µgal/d (imp)• µgal/h (imp)• µgal/min (imp)• µgal/s (imp)• mgal/d (imp)• mgal/h (imp)• mgal/min (imp)• mgal/s (imp)• Mgal/s (imp)• µbbl/d (us;oil)• µbbl/h (us;oil)• µbbl/min (us;oil)• µbbl/s (us;oil)• kImpGal• Btu/lb• oz/ft181

Description of device parameters• dB• kPa a• kPa g• MPa a• MPa g• Pa a• Pa g• g/d• g/h• g/min• g/s• kg/d• kg/h• kg/min• kg/s
• cm
• km
• nm
• pm
•a
• ha
• cm²
• dm²
• km²
• m²
• mm²
• mm³
• cl
•t
• kg/m³• kg/dm³• g/cm³• g/m³• kg/l• kgf/cm²
• GJ
• kJ
• MJ
• km/h• kt• m/s²
• GHz
• kHz
• MHz
• 1/min• 1/s• THz• rad/s• 1/s²
• Mg
• mg
• g/l
• g/ml• Mg/m³• t/m³• mg/m• tex• kg/m• kgm/s
182

Proline Prowirl 200 FOUNDATION Fieldbus• gal/d (us)• gal/h (us)• gal/min (us)• gal/s (us)• Mgal/d (us)• bbl/d (us;oil)• bbl/h (us;oil)• bbl/min (us;oil)• bbl/s (us;oil)• Mgal/h (us)• Mgal/min (us)• Mgal/s (us)• Mgal (us)• af
• af/d• af/h• af/min• af/s• bbl/d (us;beer)• bbl/h (us;beer)• bbl/min (us;beer)• bbl/s (us;beer)• kgal/d (us)• kgal/h (us)• kgal/min (us)• kgal/s (us)• µgal/d (us)• µgal/h (us)• µgal/min (us)• µgal/s (us)• mgal/d (us)• mgal/h (us)• mgal/min (us)• mgal/s (us)• Mbbl/d (us;oil)• Mbbl/h (us;oil)• Mbbl/min (us;oil)• Mbbl/s (us;oil)• mbbl/d (us;oil)• mbbl/h (us;oil)• mbbl/min (us;oil)• mbbl/s (us;oil)• kft³/d• kft³/h• kft³/min• kft³/s• mft³/d• mft³/h• mft³/min• mft³/s• kbbl(US Beer)/d• kbbl(US Beer)/h• kbbl(US Beer)/min• ubbl(US Beer)/min• ubbl(US Beer)/s• mbbl(US Beer)/d• mbbl(US Beer)/h• mbbl(US Beer)/min• mbbl(US Beer)/s
Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus• kgm²• kgm²/s
• kNm
• MNm
• mNm
• Nm
• kN
• MN
• µN
• mN
•N
• mN/m
• N/m
• atm
• GPa
• hPa
• kPa
• MPa
• µPa
• mPa
• Pa
• torr• gf/cm²
• cal
• EJ
• mJ
• PJ
• TJ
• TWh
• Wh
• GW
• µW
• mW
• nW
• pW
• TW
• Mcal/h• kcal/d• kcal/h• kcal/min• kcal/s• Mcal/d• Mcal/min• Mcal/s• kJ/d• kJ/h• kJ/min• kJ/s
•A
• mA
• kA
• µA
• nA
• pA
•C
• kC
• MC
• µC
• nC
Endress+Hauser

Description of device parameters• kgal (us)• ac-in/d• ac-in/h• ac-in/m• ac-in/s• Mft³/d• ac-in• Mft³• inH2Oa• inH2Og• inH2O a(4°C)• inH2Og(4°C)• inH2O a(68°F)• inH2Og(68°F)• ftH2Oa• ftH2Og• ftH2O a(4°C)• ftH2Og(4°C)• ftH2O a(68°F)• ftH2Og(68°F)• inHga• inHgg• inHg a(0°C)• inHgg(0°C)• klb(US)/d• klb(US)/h• klb(US)/min• klb(US)/s• MSft³/D• mils/yr• ft/s²• MLB/H• lbf-in/deg

183

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

• pC
• Ah
• W/mK• m2K/W• W/m²K• J/K• kJ/K• J/(kgK)• kJ/(kgK)• J/kg• kJ/kg• MJ/kg• C/cm³• C/m³• C/mm³• kC/m³• µC/m³• mC/m³• C/cm²• C/m²• C/mm²• kC/m²• µC/m²• mC/m²• kV/m• MV/m
• µV/m• mV/m
• V/cm
• V/m
• kV
• MV
• µV
•F
• µF
• mF
• nF
• F/m
• A/cm²• kA/m²• MA/m²• A/cm• A/m
• kA/m
• µT
• mT
• nT
•T
• mWb
• Wb
• kWb/m• Wb/m
•H
• µH
• mH
• nH
• pH
• H/m
• µH/m184

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• nH/m• Am²• GOhm
• kOhm• MOhm• µOhm• mOhm• kS• µS• µS/cm
• mS
•S
• t/d
• t/h
• t/min• t/s•%• m³/d• m³/h• m³/min• m³/s• GOhmm• kOhmm• MOhmm• µOhmm• mOhmm• nOhmm• Ohmcm• Ohmm
• kS/m• MS/m• µS/mm• mS/cm
• S/m
• sr
• l/d
• l/h
• l/min• l/s• Ml/d• kW/m²• W/(sr-m²)• cd• cd/m²• lm• lm/m²• lm/W
• lmh
• lms
• lx
• lxs
• µW/m²• mW/m²• pW/m²• Pas/m³• Pas/m• ppm• MJ/d• MJ/minEndress+Hauser

185

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• MJ/s• cm³/d• cm³/h• cm³/min• cm³/s• Nm³• Nm³/d• Nm³/h• Nm³/min• Nm³/s• Sm³• Sm³/d• Sm³/h• Sm³/min• Sm³/s• Nl• Nl/d• Nl/h• Nl/min• Nl/s• Sl• ml/min•B• ppb• ppth• °Balling• km³/d• km³/h• km³/min• km³/s• Mm³/d• Mm³/h• Mm³/min• Mm³/s• µm³/d• µm³/h• µm³/min• µm³/s• mm³/d• mm³/h• mm³/min• mm³/s• kl/d• kl/h• kl/min• kl• Sl/d• Sl/h• Sl/min• Sl/s• kL/s• Ml/h• Ml/min• Mm³/d• Mm³• GPa a• GPa g• mPa a• mPa g186

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• µPa a• µPa g• hPa a• hPa g• gf/cm² a• gf/cm² g• kgf/cm² a• kgf/cm² g• mBarg• µbar• Gy• kcal/kg
• mGy
• mSv
• rad
• rem
• Sv
• Bq
• kBq
• MBq
• cnt/s• MSft³/d• SCCM• dm• mm/yr• g/m• µg/m³• µg/l• mg/m³• kmol• µmol• mmol• mol• mol/dm³• mol/m³• mol/l• cm³/mol• dm³/mol• m³/mol• g/mol• kg/mol• l/mol• mmol/kg• mol/kg• mg/l• µS/m• mS/m• nS/cm• S/cm
• kOhmcm• MOhmcm• l/m³• L/m• µL/L• ml/m³• ml/l
• %Sat• % sol/vol• % sol/mass
Endress+Hauser

187

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• %vol
• WT-%• J/mol• kJ/mol• J/(molK)• Bq/kg• kBq/kg• MBq/kg• mV/K• V/K• J/g• mV/pH• pH• pH/°C• mV/%• %/s• %/V• nA/ppm• 1/32 ms
• 1/H
• /cm
• 1/K
• 1/m
• 1/mm
• A/hPa• A/Pa• Nm²/A• Pa/A• pA/hPa• C/kg• mC/kg• dyne-cm/deg• newton-m/deg• bar a• bar gCustom-specific units
PV/SecFactory setting

K

Additional information

ResultThe selected unit applies for:• Value parameter (→  167)
• Value parameter (→  170)
• EU at 100% parameter (→  176)
• EU at 0% parameter (→  176)
• Low Cutoff parameter (→  193)
• High High Limit parameter (→  201)
• Float Value parameter (→  205)
• High Limit parameter (→  202)
• Float Value parameter (→  207)
• Low Limit parameter (→  203)
• Float Value parameter (→  208)
• Low Low Limit parameter (→  203)
• Float Value parameter (→  210)

188

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

DecimalNavigation

Expert → Analog inputs → Analog input 1 to 4 → Decimal (6961–1 to 4)

Description

Use this function to select the number of decimal places for the input value from thetransducer block (input value).

User entry–128 to 127

Factory setting

0

Grant
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Grant (6926–1 to 4)

Description

Option for releasing certain access authorizations of the field bus host system on thedevice.

Selection• Program• Tune• Alarm• Local• Operate• Service• Diagnostic

Deny
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Deny (6925–1 to 4)

Description

Option for restricting certain access authorizations of the field bus host system on thedevice.

Selection• Program Denied• Tune Denied• Alarm Denied• Local• Operate Denied

I/O Options
NavigationDescription

Option for activating the low flow cut off.

Selection

Low Cutoff

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → I/O Options (6941–1 to 4)

189

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
The limit value for the low flow cut off is defined in Low Cutoff parameter(→  193).

ChannelNavigation

 Expert → Analog inputs → Analog input 1 to 4 → Channel (6902–1 to 4)

Description

Use this function to select the input value that should be processed in the analog input
function block.

Selection• Uninitialized• Mass flow
• Flow velocity• Volume flow
• Corrected volume flow
• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *• Reynolds number *• Totalizer 1• Totalizer 2• Totalizer 3

Factory setting

Uninitialized

Status Options
Navigation

Prerequisite

The measuring device must be in the OOS operating mode so that the parameter can beedited.

Description

Use this function to select an option for the status of the output value that is supported by
the analog input block.

Selection• Propagate Fault Fwd• Uncertain if Limited• Bad if Limited• Uncertain if Man.

*

190

Expert → Analog inputs → Analog input 1 to 4 → Status Options (6971–1 to 4)

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Options• Propagate Fault Fwd
If the measuring device has the status Bad DeviceFailure or Bad SensorFailure, thedevice continues to measure and no alarm is triggered. The use of this substatus in the
output value (OUT) is defined by Propagate Fault Fwd option. With the aid of thisoption, the user/operator can specify whether the alarm generation (sending an alarm)is triggered by the block or is forwarded downstream.
• Uncertain if Limited
If the measured or calculated value reaches a limit value, Uncertain if Limited option isused for the output status.
• Bad if Limited
If the measured value exceeds or falls below an upper or lower limit value, Bad if
Limited option is used as the output status.
• Uncertain if Man.
If the Actual mode of the function block is in the Man option operating mode, Uncertain
if Man. option is used as the output status.

Linearization Type
Navigation

 Expert → Analog inputs → Analog input 1 to 4 → Lin Type (6905–1 to 4)

Description

Use this function to select the type of linearization of the input or simulation value.

Selection• Uninitialized• Direct• Indirect• Indirect Sq Root

Factory setting

Uninitialized

Additional information"Uninitialized" option
The function block does not switch to Auto operating mode.
"Direct" option
The measured value from the transducer block (input value) bypasses the linearization
function and is routed unchanged through the analog input function block (Xd_Scale =Out_Scale). This is selected if the input value already has the required physical units.
PV = Input value
The units in Units index (→  176) (Xd_Scale) and Units index (→  180) (Out_Scale)must be the same. Otherwise, the function block will remain in the OOS operating modeand the BlockConfiguration block error is displayed in Block Error (→  165).
"Indirect" option
The measured value from the transducer block (input value) is linearly rescaled via the
Xd_Scale input scaling to the required Out_Scale output range.

Endress+Hauser

191

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

PV =X⋅ (Y - Z) -Z
100
A0024820

PV
X
YZ

Primary value
Value (→  196) (Field_Val)
EU at 100% (→  180) (Out_Scale)
EU at 0% (→  180) (Out_Scale)"Indirect Sq Root" option
The measured value from the transducer block (input value ) is rescaled via the Xd_Scaleparameter group and recalculated using a root function. It is then rescaled again to therequired output range via the Out_Scale parameter group.
PV = √X⋅ (Y - Z) -Z
100
A0024847PVX
YZ

Primary value
Value (→  196) (Field_Val)
EU at 100% (→  180) (Out_Scale)
EU at 0% (→  180) (Out_Scale)Example• System unit in transducer block: kg/h
• Sensor measuring range: 0 to 30 kg/h
• Output range for the automation system: 0 to 100 %The analog input function block must be configured as follows:1.

In Channel parameter (→  190), select Mass flow option.

2.

In Linearization Type parameter (→  191), select Indirect option.
 The "Volume Flow" process variable of the transducer block "Flow" is linearly
rescaled in the AI block to the required Out_Scale output range via the Xd_Scale
input scaling.

3.

In the Xd_Scale parameter group: EU at 0% parameter (→  176), enter the value 0.
In EU at 100% parameter (→  176), enter the value 30.
In Units index parameter (→  176), select kg/h option.

4.

In the Out_Scale parameter group: EU at 0% parameter (→  180), enter the value 0.
In EU at 100% parameter (→  180), enter the value 100.
In Units index parameter (→  180), select % option.

The result is that an input value of 15 kg/h, for example, outputs a value of 50 % via the
Value parameter (→  170).

192

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

AI
A
30

B %
100

TB
50 %

15
0

100 %

0

%
100

0…30
A0027100

8Engineering unit in kg/h

Navigation

Expert → Analog inputs → Analog input 1 to 4 → Low Cutoff (6956–1 to 4)

Description

Use this function to enter a limit value for low flow cut off.

User entry

Positive floating-point number

Factory setting0

Additional information

Description

Low Cutoff

If the converted input value (PV) falls below this limit value, then it is set to zero.
The low flow cut off is enabled via I/O Options parameter (→  189).
Dependency
The unit is taken from the: Units index parameter (→  180)

StatusNavigationDescription

Displays the status of the unprocessed measured value from the device, which reflects thestatus of the transducer before signal transmission.

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Status (6923–1 to 4)

193

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit
194

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState ConstantEndress+Hauser

195Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Value (6924–1 to 4)

Description

Displays the unprocessed measured value from the device as a % of the primary value (PV).

User interface

Signed floating-point number

Process Value Filter Time
Navigation

 Expert → Analog inputs → Analog input 1 to 4 → PV Filter Time (6909–1 to 4)

Description

Use this function to enter the filter time specification for the filtering of the unconverted
input value (PV).

User entry

Positive floating-point number

Factory setting

0s

Additional information

Factory setting
If the value 0 s is entered, filtering will not be performed.

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6978–1 to 4)

Description

Option for manually acknowledging an update of the static block parameter.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new diagnostic event occurs, the measuring device sets Unacknowledged
option.
• If the diagnostic event has been acknowledged, the user can set Acknowledged
option.

196

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Update State
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Update State (6979–1 to 4)

Description

Displays the status of an update of the static block parameter. The status indicates
whether the update was communicated or not.

User interface• Uninitialized• Reported• Not Reported

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Time Stamp (6977–1 to 4)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of an update to the static block parameter that has not yet beencommunicated was identified. The time stamp is retained until the update confirmation isreceived.

User interface

Days (d), hours (h), minutes (m), seconds (s)Static revision
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Static revision (6976–1 to 4)

Description

Displays the Static revision: Each instance of a static block parameter being accessed withwrite and communicate access due to an update is counted (event counter).

User interface

0 to 65 535

Relative Index
NavigationDescription

Displays the relative index of the static block parameter that triggered the alarm (OD index
minus field bus start index).

User interface

0 to 65 535

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Relative Index (6975–1 to 4)

197

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6920–1 to 4)

Description

Option for manually acknowledging a block alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Blk Alm Auto Ack option is not enabled in Acknowledge Option parameter(→  200), the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Alarm State (6917–1 to 4)

Description

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Time Stamp
NavigationDescription

Displays the time stamp indicating when the analysis of the block was started and when astatus change of the block alarm that has not yet been communicated to the field bus hostsystem was detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)198Expert → Analog inputs → Analog input 1 to 4 → Time Stamp (6919–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

SubcodeNavigation

Expert → Analog inputs → Analog input 1 to 4 → Subcode (6918–1 to 4)

Description

Displays the specific cause of the block alarm.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Value (6921–1 to 4)

Description

Displays the value of the affected parameter at the time at which the block alarm was
detected.

User interface

0 to 255

Current
NavigationDescription

Use this function to view the current status of the process alarms.

User interface• HiHi Alarm• Hi Alarm• LoLo Alarm• Lo Alarm• Block Alarm

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Current (6912–1 to 4)

199

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6914–1 to 4)

Description

Displays an unacknowledged process alarm.

User interface• HiHi Alarm Unack• Hi Alm Unack• LoLo Alm Unack• Lo Alm Unack• Block Alm Unack

Unreported
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unreported (6915–1 to 4)

Description

Displays a process alarm that has not been communicated.

User interface• HiHi Alm Unrep• Hi Alm Unrep• LoLo Alm Unrep• Lo Alm Unrep• Block Alm Unrep

Disabled
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Disabled (6913–1 to 4)

Description

Option for disabling a process alarm category.

Selection• HiHi Alm Disabled• Hi Alm Disabled• LoLo Alm Disabled• Lo Alm Disabled• Block Alm Disabled

Acknowledge Option
NavigationDescription

Option for automatic acknowledgment of process alarms in a specific category.

Selection• HiHi Alm Auto Ack• Hi Alm Auto Ack• LoLo Alm Auto Ack• Lo Alm Auto Ack

200

Expert → Analog inputs → Analog input 1 to 4 → Ack. Option (6910–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Blk Alm Auto Ack• Fail Alm Auto Ack• Off Spec Alm Auto Ack• Maint Alm Auto Ack• Check Alm Auto Ack
Additional information

Description
Use this function to determine whether an alarm must be acknowledged via the field bus
host system.
If the process alarm option has not been enabled in this parameter, this process alarm
must only be acknowledged in Unacknowledged parameter (→  200).
Current parameter (→  199) indicates the current status of all process alarms.

Alarm Hysteresis
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Alarm Hysteresis (6911–1 to 4)

Description

Use this function to enter the hysteresis value for the upper and lower warning or alarm
limit values.

User entry

0.000000 to 50.0000 %Factory setting0%

High High Priority
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Hi Hi Priority (6938–1 to 4)

Description

Use this function to enter the priority for the upper alarm limit, which determines the
behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

High High Limit
NavigationDescription

Use this function to enter the value for the upper alarm limit.

User entry

Signed floating-point number

Factory setting0

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → High High Limit (6937–1 to 4)

201

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Dependency
The unit is taken from the: Units index parameter (→  180)

High Priority
Navigation

Expert → Analog inputs → Analog input 1 to 4 → High Priority (6940–1 to 4)

Description

Use this function to enter the priority for the upper early warning limit, which determines
the behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

High Limit
Navigation

Expert → Analog inputs → Analog input 1 to 4 → High Limit (6939–1 to 4)

Description

Use this function to enter the value for the upper early warning limit.

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Low Priority
NavigationDescription

Use this function to enter the priority for the lower early warning limit, which determines
the behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

202

Expert → Analog inputs → Analog input 1 to 4 → Low Priority (6955–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Low Limit
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Low Limit (6947–1 to 4)

Description

Use this function to enter the value for the lower early warning limit.

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Low Low Priority
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Low Low Priority (6954–1 to 4)

Description

Use this function to enter the priority for the lower alarm limit, which determines the
behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

Low Low Limit
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Low Low Limit (6953–1 to 4)

Description

Use this function to enter the value for the lower alarm limit.

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Unacknowledged
NavigationDescription

Option for manually acknowledging a process alarm that has exceeded the upper alarm
limit (High High Limit (→  201)).

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6935–1 to 4)

203

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If HiHi Alm Auto Ack option is not enabled in Acknowledge Option parameter(→  200), the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Alarm State (6932–1 to 4)

Description

Displays the status of the process alarm for the upper alarm limit. The status indicates
whether the process alarm is active and whether it has already been communicated to thefield bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Time Stamp (6934–1 to 4)

Description

Displays the time stamp of the process alarm for the upper alarm limit. This records thetime at which analysis of the block was started and at which a status change of the process
alarm that has not yet been communicated to the field bus host system was detected. The
time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigationDescription

Displays the specific cause of the process alarm for the upper alarm limit.

User interface• Other• BlockConfiguration• LinkConfiguration

204

Expert → Analog inputs → Analog input 1 to 4 → Subcode (6933–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Float Value (6936–1 to 4)

Description

Displays the value of the affected parameter at the time at which the process alarm for theupper alarm limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6930–1 to 4)

Description

Option for manually acknowledging a process alarm that has exceeded the upper early
warning limit (High Limit (→  202)).

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Hi Alm Auto Ack option is not enabled in Acknowledge Option parameter (→  200),
the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Endress+Hauser

205Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Alarm State (6927–1 to 4)

Description

Displays the status of the process alarm for the upper early warning limit. The statusindicates whether the process alarm is active and whether it has already beencommunicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Time Stamp (6929–1 to 4)

Description

Displays the time stamp of the process alarm for the upper early warning limit. This
records the time at which analysis of the block was started and at which a status change ofthe process alarm that has not yet been communicated to the field bus host system was
detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigationDescription

Displays the specific cause of the process alarm for the upper early warning limit.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

206

Expert → Analog inputs → Analog input 1 to 4 → Subcode (6928–1 to 4)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Float Value (6931–1 to 4)

Description

Displays the value of the affected parameter at the time at which the process alarm for theupper early warning limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6945–1 to 4)

Description

Option for manually acknowledging a process alarm that has exceeded the lower early
warning limit (Low Limit (→  203)).

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Lo Alm Auto Ack option is not enabled in Acknowledge Option parameter (→  200),
the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
NavigationDescription

Displays the status of the process alarm for the lower early warning limit. The statusindicates whether the process alarm is active and whether it has already beencommunicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Alarm State (6942–1 to 4)

207

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Time Stamp (6944–1 to 4)

Description

Displays the time stamp of the process alarm for the lower early warning limit. This
records the time at which analysis of the block was started and at which a status change ofthe process alarm that has not yet been communicated to the field bus host system was
detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigation

Expert → Analog inputs → Analog input 1 to 4 → Subcode (6943–1 to 4)

Description

Displays the specific cause of the process alarm for the lower early warning limit.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Float Value (6946–1 to 4)

Description

Displays the value of the affected parameter at the time at which the process alarm for the
lower early warning limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

208

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Unacknowledged (6951–1 to 4)

Description

Option for manually acknowledging a process alarm that has exceeded the lower alarm
limit (Low Low Limit (→  203)).

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If LoLo Alm Auto Ack option is not enabled in Acknowledge Option parameter(→  200), the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Alarm State (6948–1 to 4)

Description

Displays the status of the process alarm for the lower alarm limit. The status indicates
whether the process alarm is active and whether it has already been communicated to thefield bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

Time Stamp
NavigationDescription

Displays the time stamp of the process alarm for the lower alarm limit. This records thetime at which analysis of the block was started and at which a status change of the process
alarm that has not yet been communicated to the field bus host system was detected. The
time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Expert → Analog inputs → Analog input 1 to 4 → Time Stamp (6950–1 to 4)

209

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

SubcodeNavigation

Expert → Analog inputs → Analog input 1 to 4 → Subcode (6949–1 to 4)

Description

Displays the specific cause of the process alarm for the lower alarm limit.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to 4 → Float Value (6952–1 to 4)

Description

Displays the value of the affected parameter at the time at which the process alarm for the
lower alarm limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  180)

3.6

"Discrete inputs" submenu

Navigation

 Expert → Discrete inputs‣ Discrete inputs‣ Discrete input 1 to 2210

→  211

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

3.6.1

Description of device parameters"Discrete input 1 to 2" submenu

Navigation

 Expert → Discrete inputs → Discrete input 1 to 2

‣ Discrete input 1 to 2
Block tag (6851–1 to 2)→  211

Status (6853–1 to 2)→  217

Value (6854–1 to 2)→  220

Channel (6852–1 to 2)→  228

Process Value Filter Time(6855–1 to 2)→  228

Block tag
Navigation

 Expert → Discrete inputs → Discrete input 1 to 2 → Block tag (6851–1 to 2)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Static Revision
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Static Revision (6884–1 to 2)

Description

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Endress+Hauser

211

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Tag Description
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Tag Description (6885–1 to 2)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Strategy (6883–1 to 2)

Description

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Alert Key
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Alert Key (6846–1 to 2)

Description

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
NavigationDescription

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

212

Expert → Discrete inputs → Discrete input 1 to 2 → Target mode (6873–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

OOS

Additional information

Options

Description of device parameters

Detailed description of the options available for selection: Target mode parameter(→  141)

Actual mode
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Actual mode (6870–1 to 2)

Description

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modein which the function block is currently operating. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  212).

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

User interface
Detailed description of the options available for selection: Target mode parameter(→  141)

Permitted mode
NavigationDescription

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  212) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Permitted mode (6872–1 to 2)

213

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Normal mode
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Normal mode (6871–1 to 2)

Description

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Block Error
NavigationDescription

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

214

Expert → Discrete inputs → Discrete input 1 to 2 → Block Error (6857–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

StatusNavigationDescription

Displays the status of the discrete input value (PV).

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Status (6875–1 to 2)

215

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited
216

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState Constant

Value
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Value (6876–1 to 2)

Description

Displays the discrete process variable that is used for the block implementation.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

StatusNavigation

 Expert → Discrete inputs → Discrete input 1 to 2 → Status (6853–1 to 2)

Description

Displays the status of the discrete output value.

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited

Endress+Hauser

217

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit
218

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState ConstantEndress+Hauser

219

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Value
Navigation

 Expert → Discrete inputs → Discrete input 1 to 2 → Value (6854–1 to 2)

Description

Use this function to select the discrete output value.

Selection• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

Factory setting

State 0

Simulate Status
NavigationDescription

Use this function to select the status that is used for the transducer block simulation.

Selection• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit

220

Expert → Discrete inputs → Discrete input 1 to 2 → Simulate Status (6878–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const
Endress+Hauser

221

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState Constant
Factory setting

Bad NonSpecific NotLimited

Simulate Value
NavigationDescription

Use this function to select the simulation value that is used for the transducer block
simulation.

222

Expert → Discrete inputs → Discrete input 1 to 2 → Simulate Value (6879–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

Factory setting

State 0Description of device parameters

Transducer Status
NavigationDescription

Displays the current status of the transducer block.

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Transducer Stat (6880–1 to 2)

223

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const
224

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState ConstantTransducer Value
NavigationDescription

Displays the current value of the transducer block.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Transducer Value (6881–1 to 2)

225

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• State 12• State 13• State 14• State 15• State 16
Factory setting

State 0

Simulate En/Disable
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Sim. En/Disable (6877–1 to 2)

Description

Use this function to enable or disable the function block simulation.

Selection• Uninitialized• Disabled• Active

Factory setting

Uninitialized

Transducer State
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Transducer State (6891–1 to 2)

Description

Use this function to enter the transducer state: this is required by FieldValD (Statusparameter (→  229), Value parameter (→  231)) to display the actual on/off statusof the hardware.

User entry

0 to 65 535

Factory setting0

Output State
NavigationDescription

Use this function to enter the output state: This is required for scaling the discrete input
value (PV).

User entry

0 to 65 535

Factory setting

0226

Expert → Discrete inputs → Discrete input 1 to 2 → Output State (6874–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Deny
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Deny (6867–1 to 2)

Description

Option for restricting certain access authorizations of the field bus host system on thedevice.

Selection• Program Denied• Tune Denied• Alarm Denied• Local• Operate Denied

Grant
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Grant (6868–1 to 2)

Description

Option for releasing certain access authorizations of the field bus host system on thedevice.

Selection• Program• Tune• Alarm• Local• Operate• Service• Diagnostic

I/O Options
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → I/O Options (6869–1 to 2)

Description

Option for activating the inversion of the signal.

Selection

InvertAdditional information

Description
This selection determines whether the discrete input is inverted before it is stored as aprocess variable. Normally a discrete value of 0 is regarded as a logical value of 0 and adiscrete value that is not equal to 0 is regarded as a logical value of 1. If inversion isenabled, this correlation is reversed. A field device input value that is not equal to 0 results
in a discrete output value of 0 and an input value of 0 results in a discrete output value of1.

Endress+Hauser

227

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Status Options
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Status Options (6882–1 to 2)

Prerequisite

The measuring device must be in the OOS operating mode so that the parameter can beedited.

Description

For selecting an option for the status of the output value that is supported by the Discrete
input block.

Selection• Propagate Fault Fwd• Uncertain if Man.

Additional information

Options• Propagate Fault Fwd
If the measuring device indicates the status Bad DeviceFailure or Bad SensorFailure,
the device continues to measure and no alarm is triggered. The use of this substatus in
the output value (OUT) is defined by Propagate Fault Fwd option. With the aid of thisoption, the user/operator can specify whether the alarm generation (sending an alarm)is triggered by the block or is forwarded downstream.
• Uncertain if Man.
If the Actual mode of the function block is in the Man option operating mode, Uncertain
if Man. option is used as the output status.

ChannelNavigation

 Expert → Discrete inputs → Discrete input 1 to 2 → Channel (6852–1 to 2)

Description

Use this function to select the input value that should be processed in the discrete input
function block.

Selection• Uninitialized• Low flow cut off• Switch output status• Verification status

Factory setting

Switch output status

Process Value Filter Time
Navigation

 Expert → Discrete inputs → Discrete input 1 to 2 → PV Filter Time (6855–1 to 2)

Description

Use this function to enter the filter time specification for the filtering of the unconverted
input value (PV).

User entry

Positive floating-point number

Factory setting

0s

228

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Factory setting
If the value 0 s is entered, filtering will not be performed.

StatusNavigationDescription

Displays the status of the discrete input value from a measuring device in the field.

User interface• Bad NonSpecific NotLimited• Bad NonSpecific LowLimited• Bad NonSpecific HighLimited• Bad NonSpecific Constant• Bad ConfigurationError NotLimited• Bad ConfigurationError LowLimited• Bad ConfigurationError HighLimited• Bad ConfigurationError Constant• Bad NotConnected NotLimited• Bad NotConnected LowLimited• Bad NotConnected HighLimited• Bad NotConnected Constant• Bad DeviceFailure NotLimited• Bad DeviceFailure LowLimited• Bad DeviceFailure HighLimited• Bad DeviceFailure Constant• Bad SensorFailure NotLimited• Bad SensorFailure LowLimited• Bad SensorFailure HighLimited• Bad SensorFailure Constant• Bad NoComm WithLastUsableValue NotLimit• Bad No Com WithLastUsableValue LowLim• Bad No Com WithLastUsableValue HighLim• Bad No Com WithLastUsableValue Const• Bad NoComm WithNoUsableValue NotLimit• Bad NoComm WithNoUsableValue LowLimit• BadNoComm WithNoUsableValue HighLimit• Bad NoComm WithNoUsableValue Constant• Bad OutOfService NotLimited• Bad OutOfService LowLimited• Bad OutOfService HighLimited• Bad OutOfService Constant• Uncertain NonSpecific NotLimited• Uncertain NonSpecific LowLimited• Uncertain NonSpecific HighLimited• Uncertain NonSpecific Constant• Uncertain LastUsableValue NotLimited• Uncertain LastUsableValue LowLimited• Uncertain LastUsableValue HighLimited• Uncertain LastUsableValue Constant• Uncertain SubstituteValue NotLimited• Uncertain SubstituteValue LowLimited• Uncertain SubstituteValue HighLimited• Uncertain SubstituteValue Constant• Uncertain InitialValue NotLimited

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Status (6865–1 to 2)229

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Uncertain InitialValue LowLimited• Uncertain InitialValue HighLimited• Uncertain InitialValue Constant• Uncertain SensConvNotAccurate NoLimit• Uncertain SensConvNotAccurate LowLimit• Uncertain SensConvNotAccurat HighLimit• Uncertain SensConvNotAccurat Const• Uncertain EngUnitRangeViolation NotLimit• Uncertain EngUnitRangeViolation LowLimit• Uncertain EngUnitRangeViolation HighLim• Uncertain EngUnitRangeViolation Const• Uncertain Subnormal NotLimited• Uncertain Subnormal LowLimited• Uncertain Subnormal HighLimited• Uncertain Subnormal Constant• Good NonCAS NonSpecific NotLimited• Good NonCAS NonSpecific LowLimited• Good NonCAS NonSpecific HighLimited• Good NonCAS NonSpecific Constant• Good NonCAS ActiveBlockAlarm NotLimit• Good NonCAS ActiveBlockAlarm LowLimit• Good NonCAS ActiveBlockAlarm HighLimit• Good NonCAS ActiveBlockAlarm Const• Good NonCAS ActiveAdvisoryAlarm NotLimit• Good NonCAS ActiveAdvisoryAlarm LowLimit• Good NonCAS ActAdvisoryAlarm HighLimit• Good NonCAS ActAdvisoryAlarm Const• Good NonCAS ActiveCriticalAlarm NotLimit• Good NonCAS ActiveCriticalAlarm LowLimit• Good NonCAS ActiveCriticalAlarm HL• Good NonCAS ActiveCriticalAlarm Con.
• Good NonCAS UnackBlockAlarm NotLimit• Good NonCAS UnackBlockAlarm LowLimit• Good NonCAS UnackBlockAlarm HighLimit• Good NonCAS UnackBlockAlarm Const• Good NonCAS UnackAdvisoryAlarm NotLimit• Good NonCAS UnackAdvisoryAlarm LowLimit• Good NonCAS UnackAdvisoryAlarm HighLimit• Good NonCAS UnackAdvisoryAlarm Const• Good NonCAS UnackCriticalAlarm NotLimit• Good NonCAS UnackCriticalAlarm LowLimit• Good NonCAS UnackCriticalAlarm HighLimit• Good NonCAS UnackCriticalAlarm Const• Good Cascade NonSpecific NotLimited• Good Cascade NonSpecific LowLimited• Good Cascade NonSpecific HighLimited• Good Cascade NonSpecific Constant• Good Cascade InitAcknowledge NotLimit• Good Cascade InitAcknowledge LowLimit• Good Cascade InitAcknowledge HighLimit• Good Cascade InitAcknowledge Const• Good Cascade InitRequest NotLimit• Good Cascade InitRequest LowLimit• Good Cascade InitRequest HighLimit• Good Cascade InitRequest Const• Good Cascade NotInvited NotLimited• Good Cascade NotInvited LowLimited• Good Cascade NotInvited HighLimited• Good Cascade NotInvited Constant
230

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Good Cascade NotSelected NotLimited• Good Cascade NotSelected LowLimited• Good Cascade NotSelected HighLimited• Good Cascade NotSelected Constant• Good Cascade DoNotSelect NotLimited• Good Cascade DoNotSelect LowLimited• Good Cascade DoNotSelect HighLimited• Good Cascade DoNotSelect Constant• Good Cascade LocalOverride NotLimited• Good Cascade LocalOverride LowLimited• Good Cascade LocalOverride HighLimited• Good Cascade LocalOverride Constant• Good Cascade FaultStateActive NotLimit• Good Cascade FaultStateActive LowLimit• Good Cascade FaultStateActive HighLimit• Good Cascade FaultStateActive Constant• Good Cascade InitFaultState NotLimited• Good Cascade InitFaultState LowLimited• Good Cascade InitFaultState HighLimited• Good Cascade InitiateFaultState Constant
Factory setting

Bad NonSpecific NotLimited

Additional information

Description
An output parameter can be linked to an input parameter from another function block.
Both the input parameter and the output parameter have a field value and status. The
status of the input parameter is taken from the linked output parameter.

Value
NavigationDescription

Displays the discrete input value from a measuring device in the field.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

Factory setting

State 0Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Value (6866–1 to 2)

231

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Unacknowledged
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Unacknowledged (6889–1 to 2)

Description

Option for manually acknowledging an update of the static block parameter.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new diagnostic event occurs, the measuring device sets Unacknowledged
option.
• If the diagnostic event has been acknowledged, the user can set Acknowledged
option.

Update State
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Update State (6890–1 to 2)

Description

Displays the status of an update of the static block parameter. The status indicates
whether the update was communicated or not.

User interface• Uninitialized• Reported• Not Reported

Time Stamp
NavigationDescription

Displays the time stamp indicating when the analysis of the block was started and when astatus change of an update to the static block parameter that has not yet beencommunicated was identified. The time stamp is retained until the update confirmation isreceived.

User interface

Days (d), hours (h), minutes (m), seconds (s)232Expert → Discrete inputs → Discrete input 1 to 2 → Time Stamp (6888–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Static revision
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Static revision (6887–1 to 2)

Description

Displays the Static revision: Each instance of a static block parameter being accessed withwrite and communicate access due to an update is counted (event counter).

User interface

0 to 65 535

Relative Index
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Relative Index (6886–1 to 2)

Description

Displays the relative index of the static block parameter that triggered the alarm (OD index
minus field bus start index).

User interface

0 to 65 535

Acknowledge Option
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Ack. Option (6841–1 to 2)

Description

Option for automatic acknowledgment of process alarms in a specific category.

Selection• Disc Alm Auto Ack• Blk Alm Auto Ack• Fail Alm Auto Ack• Off Spec Alm Auto Ack• Maint Alm Auto Ack• Check Alm Auto Ack

Additional information

Description
Use this function to determine whether an alarm must be acknowledged via the field bus
host system.
If the process alarm option has not been enabled in this parameter, this process alarm
must only be acknowledged in Unacknowledged parameter (→  200).
Current parameter (→  199) indicates the current status of all process alarms.

Current
NavigationDescription

Use this function to view the current status of the process alarms.

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Current (6842–1 to 2)

233

Description of device parameters

User interface

Proline Prowirl 200 FOUNDATION Fieldbus• Discrete Alarm• Block Alarm• Fail Alarm• Off Spec Alarm• Maintenance Alarm• Check Alarm

Disabled
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Disabled (6843–1 to 2)

Description

Option for disabling a process alarm category.

Selection• Disc Alm Disabled• Block Alm Disabled• Fail Alm Disabled• Off Spec Alm Disabled• Maint Alm Disabled• Check Alm Disabled

Unacknowledged
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Unacknowledged (6844–1 to 2)

Description

Displays an unacknowledged process alarm.

User interface• Disc Alm Unack• Block Alm Unack• Fail Alm Unack• Off Spec Alm Unack• Maint Alm Unack• Check Alm Unack

Unreported
NavigationDescription

Displays a process alarm that has not been communicated.

User interface• Disc Alm Unrep• Block Alm Unrep• Fail Alm Unrep• Off Spec Alm Unrep• Maint Alm Unrep• Check Alm Unrep

234

Expert → Discrete inputs → Discrete input 1 to 2 → Unreported (6845–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Alarm State
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Alarm State (6847–1 to 2)

Description

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

SubcodeNavigation

Expert → Discrete inputs → Discrete input 1 to 2 → Subcode (6848–1 to 2)

Description

Displays the specific cause of the block alarm.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Time Stamp
NavigationDescription

Displays the time stamp indicating when the analysis of the block was started and when astatus change of the block alarm that has not yet been communicated to the field bus hostsystem was detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to 2 → Time Stamp (6849–1 to 2)

235

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Unacknowledged
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Unacknowledged (6850–1 to 2)

Description

Option for manually acknowledging a block alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Blk Alm Auto Ack option is not enabled in Acknowledge Option parameter(→  233), the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Value
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Value (6856–1 to 2)

Description

Displays the value of the affected parameter at the time at which the block alarm was
detected.

User interface

0 to 255

Alarm State
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Alarm State (6858–1 to 2)

Description

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• Clear-Not Reported• Active-Reported• Active-Not Reported

SubcodeNavigationDescription

Displays the specific cause of the discrete alarm.

236

Expert → Discrete inputs → Discrete input 1 to 2 → Subcode (6859–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User interface

Description of device parameters• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

Time Stamp
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Time Stamp (6860–1 to 2)

Description

Displays the time stamp indicating when the analysis of the function block was started andwhen a status change of the discrete alarm that has not yet been communicated to thefield bus host system was detected. The time stamp is retained until the alarmconfirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Unacknowledged
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Unacknowledged (6861–1 to 2)

Description

Option for manually acknowledging a discrete alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Disc Alm Auto Ack option is not enabled in Acknowledge Option parameter(→  233), the process alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Endress+Hauser

237

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Discrete Value
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Discrete Value (6862–1 to 2)

Description

Displays the value of the associated parameter at the time at which the alarm was
detected.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

Discrete Limit
Navigation

Expert → Discrete inputs → Discrete input 1 to 2 → Discrete Limit (6863–1 to 2)

Description

Use this to enter the status of the discrete input value that triggers an alarm.

User entry

0 to 255

Factory setting

0

Discrete Priority
NavigationDescription

Use this to enter the priority of a discrete alarm.

User entry

0 to 15

Factory setting

0

238

Expert → Discrete inputs → Discrete input 1 to 2 → Discrete Prio (6864–1 to 2)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

3.7

Description of device parameters"Analog outputs" submenu

Navigation

 Expert → Analog outputs‣ Analog outputs‣ Multiple analog output

3.7.1
Navigation→  239"Multiple analog output" submenu
 Expert → Analog outputs → Multiple AO‣ Multiple analog output
Block tag (11351)→  239

Status Options (11363)→  243

Fault State Time (11354)→  244

Fault State Val 1 (11355)→  244

Fault State Val 2 (11356)→  245

Fault State Val 3 (11357)→  245

Fault State Val 4 (11358)→  245

Fault State Val 5 (11359)→  246

Fault State Val 6 (11360)→  246

Fault State Val 7 (11361)→  246

Fault State Val 8 (11362)→  247

Fault State Status (11353)→  247

Block tag
Navigation

 Expert → Analog outputs → Multiple AO → Block tag (11351)

Description

Use this function to enter the Block tag: specify a "label" for identifying the function block.

User entry

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Endress+Hauser

239

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Static Revision
Navigation

Expert → Analog outputs → Multiple AO → Static Revision (11371)

Description

Displays the Static Revision: every write access to a static block parameter is counted(event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Analog outputs → Multiple AO → Tag Description (11372)

Description

Use this function to enter the Tag Description: define a user-specific text for the detailed
description of the function block.

User entry

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Analog outputs → Multiple AO → Strategy (11370)

Description

Use this function to enter the Strategy: makes it possible to group blocks by entering
identical numbers.

User entry

0 to FFFF

Factory setting0

Alert Key
NavigationDescription

Use this function to enter the Alert Key: identifies the section of the plant where thetransmitter is located. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

240

Expert → Analog outputs → Multiple AO → Alert Key (11365)

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Target mode
Navigation

Expert → Analog outputs → Multiple AO → Target mode (11369)

Description

Use this function to select the Target mode: the selection specifies which operating modeis used for this block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  141)

Actual mode
Navigation

Expert → Analog outputs → Multiple AO → Actual mode (11366)

Description

Displays the Actual mode: under certain conditions a block may not work in the operating
mode to be used. In this case, the Actual mode represents the valid mode in which theblock is currently operating. A comparison of the Actual mode with the Target modeindicates whether the Target mode (→  241) could be reached.

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  141)

Endress+Hauser

241

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Permitted mode
Navigation

Expert → Analog outputs → Multiple AO → Permitted mode (11368)

Description

Use this function to select the Permitted mode: the selection defines which operating
modes are available for the function block in the Target mode (→  241). The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  141)

Normal mode
Navigation

Expert → Analog outputs → Multiple AO → Normal mode (11367)

Description

Use this function to select the Normal mode: this mode is provided to enable the user toselect the Normal mode among the operating modes available. This can be set using an
operating tool in order to help the user configure the operating mode of a block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

Auto

Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  141)242

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Block Error
Navigation

Expert → Analog outputs → Multiple AO → Block Error (11364)

Description

Displays the short text for the Block Error that occurred in the function block.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

ChannelNavigation

Expert → Analog outputs → Multiple AO → Channel (11352)

Description

Use this function to select the assignment or connection between the Analog Output
function block and the Transducer Block.

Selection• Uninitialized• Channel_0

Factory setting

Channel_0

Status Options
Navigation

 Expert → Analog outputs → Multiple AO → Status Options (11363)

Description

Option for determining the fault state of the function block.

Selection• Faultstate to value 1• Faultstate to value 2• Faultstate to value 3• Faultstate to value 4• Faultstate to value 5• Faultstate to value 6• Faultstate to value 7• Faultstate to value 8• Use faultstate value on restart 1

Endress+Hauser

243Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Use faultstate value on restart 2• Use faultstate value on restart 3• Use faultstate value on restart 4• Use faultstate value on restart 5• Use faultstate value on restart 6• Use faultstate value on restart 7• Use faultstate value on restart 8
Additional information

Description
This behavior is activated if an error condition of the set point that applies is present for
longer than defined in the Fault State Time parameter (→  244) or if the Set Fault
State parameter (→  152) is activated in the Resource block.
The fault state is defined via the following parameters:• Fault State Time (→  244)
• Fault State Val 1 to 8
Selection• Faultstate to value 1…8
The value predefined in the Fault State Val 1 to 8 parameter is used instead of theanalog set point. The fault state is enabled if there is an error condition.
• Use faultstate value on restart 1…8The value predefined in the Fault State Val 1 to 8 parameter is used if the device isrestarted. The non-volatile value is used otherwise. The fault state is not enabled andonly the predefined value is used.

Fault State Time
Navigation

 Expert → Analog outputs → Multiple AO → Fault State Time (11354)

Description

Use this function to enter a time range during which an error condition (of the currently
valid set point) must be met without interruption before an error message is generated.

User entry

Positive floating-point number

Factory setting

0s

Fault State Val 1
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 1 (11355)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 1 is present.

User entry

Signed floating-point number

Factory setting

0

244

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
This value is ignored if the Faultstate to value 1 option is not selected in the Status
Options parameter (→  243).

Fault State Val 2
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 2 (11356)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 2 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Faultstate to value 2 option is not selected in the Status
Options parameter (→  243).

Fault State Val 3
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 3 (11357)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 3 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Faultstate to value 3 option is not selected in the Status
Options parameter (→  243).

Fault State Val 4
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 4 (11358)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 4 is present.

User entry

Signed floating-point number

Factory setting0

Endress+Hauser

245

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
This value is ignored if the Faultstate to value 4 option is not selected in the Status
Options parameter (→  243).

Fault State Val 5
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 5 (11359)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 5 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Faultstate to value 5 option is not selected in the Status
Options parameter (→  243).

Fault State Val 6
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 6 (11360)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 6 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Faultstate to value 6 option is not selected in the Status
Options parameter (→  243).

Fault State Val 7
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 7 (11361)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 7 is present.

User entry

Signed floating-point number

Factory setting0

246

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
This value is ignored if the Faultstate to value 7 option is not selected in the Status
Options parameter (→  243).

Fault State Val 8
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 8 (11362)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 8 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Faultstate to value 8 option is not selected in the Status
Options parameter (→  243).

Fault State Status
Navigation

 Expert → Analog outputs → Multiple AO → FaultStateStatus (11353)

Description

Displays the values for which the fault state is enabled.

User interface• Value 1 is in Fault State• Value 2 is in Fault State• Value 3 is in Fault State• Value 4 is in Fault State• Value 5 is in Fault State• Value 6 is in Fault State• Value 7 is in Fault State• Value 8 is in Fault State

3.8"Discrete outputs" submenu

The Discrete outputs function block (DO, discrete output) processes a discrete set point
received from an upstream function block or a higher-level process control system thatenables various device functions (e.g. zero point adjustment or reset of the totalizer) to be
triggered in the downstream transducer block.

Endress+Hauser

247

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Navigation

 Expert → Discrete outputs‣ Discrete outputs‣ Multiple discrete output

3.8.1
Navigation→  248"Multiple discrete output" submenu
 Expert → Discrete outputs → Multiple DO‣ Multiple discrete output
Block tag (11252)→  248

Status Options (11268)→  252

Fault State Time (11255)→  253

Fault State Value Discrete 1 (11256)→  253

Fault State Value Discrete 2 (11257)→  254

Fault State Value Discrete 3 (11258)→  254

Fault State Value Discrete 4 (11259)→  254

Fault State Value Discrete 5 (11260)→  255

Fault State Value Discrete 6 (11261)→  255

Fault State Value Discrete 7 (11262)→  255

Fault State Value Discrete 8 (11263)→  256

Fault State Status (11254)→  256

Block tag
Navigation

 Expert → Discrete outputs → Multiple DO → Block tag (11252)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

248

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Static Revision
Navigation

Expert → Discrete outputs → Multiple DO → Static Revision (11270)

Description

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Discrete outputs → Multiple DO → Tag Description (11271)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Discrete outputs → Multiple DO → Strategy (11269)

Description

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Alert Key
NavigationDescription

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting1

Endress+Hauser

Expert → Discrete outputs → Multiple DO → Alert Key (11251)

249

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Target mode
Navigation

Expert → Discrete outputs → Multiple DO → Target mode (11267)

Description

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Actual mode
Navigation

Expert → Discrete outputs → Multiple DO → Actual mode (11264)

Description

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modethat the function block is currently operating in. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  250).

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

User interface
Detailed description of the options available for selection: Target mode parameter(→  141)

250

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Permitted mode
Navigation

Expert → Discrete outputs → Multiple DO → Permitted mode (11266)

Description

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  250) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Normal mode
Navigation

Expert → Discrete outputs → Multiple DO → Normal mode (11265)

Description

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

Auto

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  141)

Endress+Hauser

251Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Block Error
Navigation

Expert → Discrete outputs → Multiple DO → Block Error (11272)

Description

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfiguration• LinkConfiguration• SimulationActive• LocalOverride• DeviceFaultState• DeviceMaintenance• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeeded• PowerUp• OutOfService

ChannelNavigation

Expert → Discrete outputs → Multiple DO → Channel (11253)

Description

Option for the assignment or connection between the discrete output function block andthe transducer block.

Selection• Uninitialized• Channel_D0

Factory setting

Channel_D0

Status Options
Navigation

 Expert → Discrete outputs → Multiple DO → Status Options (11268)

Description

Option for determining the fault state of the function block.

Selection• Faultstate to value 1• Faultstate to value 2• Faultstate to value 3• Faultstate to value 4• Faultstate to value 5• Faultstate to value 6• Faultstate to value 7• Faultstate to value 8• Use faultstate value on restart 1

252

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Use faultstate value on restart 2• Use faultstate value on restart 3• Use faultstate value on restart 4• Use faultstate value on restart 5• Use faultstate value on restart 6• Use faultstate value on restart 7• Use faultstate value on restart 8
Additional information

Description
This behavior is enabled if an error condition of the valid set point persists for longer than
the time specified in Fault State Time parameter (→  253) or if Set Fault Stateparameter (→  152) is enabled in Resource block.
The fault state is defined via the following parameters:• Fault State Time (→  253)
• Fault State Value Discrete 1 to 8
Options• Faultstate to value 1…8
The value predefined in Fault State Value Discrete 1 to 8 parameter is used in place ofthe discrete set point. The fault state is enabled if there is an error condition.
• Use faultstate value on restart 1…8The value predefined in Fault State Value Discrete 1 to 8 parameter is used if thedevice is restarted. The non-volatile value is used otherwise. The fault state is notenabled and only the predefined value is used.

Fault State Time
Navigation

 Expert → Discrete outputs → Multiple DO → Fault State Time (11255)

Description

Use this function to enter a time range during which an error condition (of the currently
valid set point) must be met without interruption before an error message is generated.

User entry

Positive floating-point number

Factory setting

0s

Fault State Value Discrete 1
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 1 (11256)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 1 is present.

User entry

0 to 255

Factory setting0

Endress+Hauser

253

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
If Faultstate to value 1 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 2
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 2 (11257)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 2 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Faultstate to value 2 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 3
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 3 (11258)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 3 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Faultstate to value 3 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 4
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 4 (11259)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 4 is present.

User entry

0 to 255

Factory setting

0254

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
If Faultstate to value 4 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 5
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 5 (11260)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 5 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Faultstate to value 5 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 6
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 6 (11261)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 6 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Faultstate to value 6 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 7
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 7 (11262)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 7 is present.

User entry

0 to 255

Factory setting0

Endress+Hauser

255

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
If Faultstate to value 7 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Value Discrete 8
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 8 (11263)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 8 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Faultstate to value 8 option is not selected in Status Options parameter(→  252), this value is ignored.

Fault State Status
Navigation

 Expert → Discrete outputs → Multiple DO → FaultStateStatus (11254)

Description

Displays the values for which the fault state is enabled.

User interface• Value 1 is in Fault State• Value 2 is in Fault State• Value 3 is in Fault State• Value 4 is in Fault State• Value 5 is in Fault State• Value 6 is in Fault State• Value 7 is in Fault State• Value 8 is in Fault State

3.9"Application" submenu

Navigation

 Expert → Application‣ Application

256

Reset all totalizers (2806)→  257‣ Totalizer 1 to 3

→  257

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Reset all totalizers
Navigation

 Expert → Application → Reset all tot. (2806)

Description

Use this function to reset all totalizers to the value 0 and restart the totaling process. Thisdeletes all the flow values previously totalized.

Selection• Cancel• Reset + totalize

Factory setting

CancelAdditional information

Selection• CancelNo action is executed and the user exits the parameter.
• Reset + totalize
All totalizers are reset to 0 and the totaling process is restarted.

3.9.1"Totalizer 1 to 3" submenu

Navigation

 Expert → Application → Totalizer 1 to 3

‣ Totalizer 1 to 3
Assign process variable (0914–1 to 3)→  257

Unit totalizer (0915–1 to 3)→  258

Control Totalizer 1 to 3 (0912–1 to 3)→  260

Preset value 1 to 3 (0913–1 to 3)→  260

Failure mode (0901–1 to 3)→  261

Assign process variable

Navigation

 Expert → Application → Totalizer 1 to 3 → Assign variable (0914–1 to 3)

Description

Use this function to select a process variable for the Totalizer 1 to 3.

Selection• Off• Volume flow
• Corrected volume flow
• Mass flow

Endress+Hauser

257

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *Factory setting• Totalizer 1: Volume flow
• Totalizer 2: Mass flow
• Totalizer 3: Corrected volume flow

Additional information

Description
If the option selected is changed, the device resets the totalizer to 0.
Selection
If the Off option is selected, only Assign process variable parameter (→  257) is still
displayed in the Totalizer 1 to 3 submenu. All other parameters in the submenu arehidden.

Unit totalizer



Navigation

 Expert → Application → Totalizer 1 to 3 → Unit totalizer (0915–1 to 3)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  257) of the Totalizer 1 to 3 submenu:• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *Description

Use this function to select the unit for the process variable of totalizer 1-3.

Selection

SI units
•g
• kg
•t

US units• oz• lb• STonor*

258Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega

Description of device parameters

US units• af• ft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;liq.)• bbl (us;beer)• bbl (us;oil)• bbl (us;tank)

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

US units• Sft³• Sgal (us)• Sbbl (us;liq.)

Imperial units
Sgal (imp)orSI units
• Nl
• Nm³
• Sl
• Sm³
or
SI units
• kWh
• MWh
• GWh
• kJ
• MJ
• GJ
• kcal• Mcal• GcalFactory setting

Country-specific:
• m³
• ft³

Additional information

Description

Imperial units• Btu• MBtu• MMBtuThe unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  69).
Selection
The selection depends on the process variable selected in the Assign process variable
parameter (→  257).

Endress+Hauser

259

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Control Totalizer 1 to 3
Navigation

 Expert → Application → Totalizer 1 to 3 → Control Tot. 1 to 3 (0912–1 to 3)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  257) of the Totalizer 1 to 3 submenu:• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *Description

Use this function to select the control of totalizer value 1-3.

Selection• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• Preset + totalize

Factory setting

Totalize

Additional information

Selection• Totalize
The totalizer is started or continues totalizing with the current counter reading.
• Reset + hold
The totaling process is stopped and the totalizer is reset to 0.
• Preset + hold
The totaling process is stopped and the totalizer is set to its defined start value from the
Preset value parameter (→  260).
• Reset + totalize
The totalizer is reset to 0 and the totaling process is restarted.
• Preset + totalize
The totalizer is set to the defined start value in Preset value parameter (→  260) andthe totaling process is restarted.

Preset value 1 to 3
Navigation

 Expert → Application → Totalizer 1 to 3 → Preset value 1 to 3 (0913–1 to 3)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  257) of the Totalizer 1 to 3 submenu:• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow **260

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Condensate mass flow *• Energy flow *• Heat flow difference *Description

Use this function to enter a start value for totalizer 1-3.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 m³• 0 ft³

Additional information

User entry
The unit of the selected process variable is specified for the totalizer in the Unit
totalizer parameter (→  258).
ExampleThis configuration is suitable for applications such as iterative filling processes with a fixed
batch quantity.

Failure modeNavigation

 Expert → Application → Totalizer 1 to 3 → Failure mode (0901–1 to 3)

Prerequisite

One of the following options is selected in the Assign process variable parameter(→  257) of the Totalizer 1 to 3 submenu:• Volume flow
• Corrected volume flow
• Mass flow
• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *Description

Use this function to select how a totalizer behaves in an alarm condition.

Selection• Stop• Actual value• Last valid value

Factory setting

Stop

Additional information

Description
This setting does not affect the error response mode of other totalizers and theoutputs. This is specified in separate parameters.

*

Visibility depends on order options or device settings

Endress+Hauser

261

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Selection• StopTotalizing is stopped in an alarm condition.
• Actual value
The totalizer continues to count based on the actual measured value; the error is
ignored.
• Last valid value
The totalizer continues to count based on the last valid measured value before the error
occurred.

3.10"Diagnostics" submenu

Navigation

 Expert → Diagnostics‣ Diagnostics
Actual diagnostics (0691)→  262

Previous diagnostics (0690)→  263

Operating time from restart (0653)→  264

Operating time (0652)→  264‣ Diagnostic list

→  265‣ Event logbook→  268‣ Device information→  271‣ Sensor information→  275‣ Data logging→  275‣ Min/max values→  281‣ Heartbeat→  287‣ Simulation→  288

Actual diagnostics
Navigation

 Expert → Diagnostics → Actual diagnos. (0691)

Prerequisite

A diagnostic event has occurred.

262

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Description

Displays the current diagnostic message. If two or more messages occur simultaneously,
the message with the highest priority is shown on the display.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Additional pending diagnostic messages can be displayed in the Diagnostic list
submenu (→  265).
Information on what is causing the diagnostic message, and remedy measures, can beviewed by pressing the -button.
ExampleFor the display format:
F271 Main electronic failure

Timestamp
Navigation

Expert → Diagnostics → Timestamp (0667)

Description

Displays the operating time when the current diagnostic message occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Actual diagnostics parameter(→  262).
ExampleFor the display format:
24d12h13m00s

Previous diagnostics
Navigation

 Expert → Diagnostics → Prev.diagnostics (0690)

Prerequisite

Two diagnostic events have already occurred.

Description

Displays the diagnostic message that occurred before the current message.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Information on what is causing the diagnostic message, and remedy measures, can beviewed by pressing the -button.
ExampleFor the display format:
F271 Main electronic failure

Endress+Hauser

263

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Timestamp
Navigation

Expert → Diagnostics → Timestamp (0672)

Description

Displays the operating time when the last diagnostic message before the current message
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Previous diagnostics parameter(→  263).
ExampleFor the display format:
24d12h13m00s

Operating time from restart
Navigation

 Expert → Diagnostics → Time fr. restart (0653)

Description

Use this function to display the time the device has been in operation since the last device
restart.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Operating time
Navigation

 Expert → Diagnostics → Operating time (0652)

Description

Use this function to display the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The maximum number of days is 9999, which is equivalent to 27 years.

264

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

3.10.1

Description of device parameters"Diagnostic list" submenu

Navigation

 Expert → Diagnostics → Diagnostic list

‣ Diagnostic list
Diagnostics 1 (0692)→  265

Diagnostics 2 (0693)→  266

Diagnostics 3 (0694)→  266

Diagnostics 4 (0695)→  267

Diagnostics 5 (0696)→  268

Diagnostics 1
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 1 (0692)

Description

Use this function to display the current diagnostics message with the highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Examples
For the display format:• S442 Frequency output• F276 I/O module failure

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp (0683)

Description

Displays the operating time when the diagnostic message with the highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Diagnostics 1 parameter(→  265).
ExampleFor the display format:
24d12h13m00s

Endress+Hauser

265Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Diagnostics 2
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 2 (0693)

Description

Use this function to display the current diagnostics message with the second-highest
priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Examples
For the display format:• S442 Frequency output• F276 I/O module failure

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp (0684)

Description

Displays the operating time when the diagnostic message with the second-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Diagnostics 2 parameter(→  266).
ExampleFor the display format:
24d12h13m00s

Diagnostics 3
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 3 (0694)

Description

Use this function to display the current diagnostics message with the third-highest
priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Examples
For the display format:• S442 Frequency output• F276 I/O module failure

266

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp (0685)

Description

Displays the operating time when the diagnostic message with the third-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Diagnostics 3 parameter(→  266).
ExampleFor the display format:
24d12h13m00s

Diagnostics 4
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 4 (0695)

Description

Use this function to display the current diagnostics message with the fourth-highest
priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Examples
For the display format:• S442 Frequency output• F276 I/O module failure

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp (0686)

Description

Displays the operating time when the diagnostic message with the fourth-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Diagnostics 4 parameter(→  267).
ExampleFor the display format:
24d12h13m00s

Endress+Hauser

267

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Diagnostics 5
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 5 (0696)

Description

Use this function to display the current diagnostics message with the fifth-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Examples
For the display format:• S442 Frequency output• F276 I/O module failure

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp (0687)

Description

Displays the operating time when the diagnostic message with the fifth-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be displayed via the Diagnostics 5 parameter(→  268).
ExampleFor the display format:
24d12h13m00s

3.10.2"Event logbook" submenu

Navigation

Expert → Diagnostics → Event logbook‣ Event logbook

268

Filter options (0705)→  269‣ Event list

→  270

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Filter options

Navigation

Expert → Diagnostics → Event logbook → Filter options (0705)

Description

Use this function to select the category whose event messages are displayed in the event
list of the local display.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)Factory setting

All

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required

Filter options

Navigation

Expert → Diagnostics → Event logbook → Filter options (0656)

Description

Use this function to select the category whose event messages are displayed in the event
list of the operating tool.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)Factory setting

All

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required

Endress+Hauser

269

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus"Event list" submenu
Navigation

 Expert → Diagnostics → Event logbook → Event list

▸ Event list
→  270

Event list

Event list
Navigation

 Expert → Diagnostics → Event logbook → Event list

Description

Use this function to display the history of event messages of the category selected in theFilter options parameter (→  269).

User interface• For a "Category I" event message
Information event, short message, symbol for event recording and operating time whenerror occurred• For a "Category F, C, S, M" event message (status signal)
Diagnostics code, short message, symbol for event recording and operating time whenerror occurred

Additional information

Description
A maximum of 20 event messages are displayed in chronological order. If the advanced
HistoROM function is enabled in the device, the event list can contain up to 100 entries.
The following symbols indicate whether an event has occurred or has ended:• : Occurrence of the event• : End of the event
Examples
For the display format:• I1091 Configuration modified
 24d12h13m00s• S442 Frequency output
 01d04h12min30s
Additional information, such as remedial measures, can be retrieved via the  key.
HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.
To order the HistoROM advanced capabilities application package, see the"Accessories" section of the "Technical Information" document.

270

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

3.10.3

Description of device parameters"Device information" submenu

Navigation

 Expert → Diagnostics → Device info

‣ Device information
Device tag (10799)→  271

Serial number (10798)→  272

Device type (10711)→  272

Device Revision (10710)→  272

DD Revision (10709)→  272

Firmware version (10792)→  273

Hardware revision (10793)→  273

ITK Version (10794)→  273

Extended order code (10795)→  273

Extended order code 1 (10796)→  274

Extended order code 2 (10797)→  274

ENP version (10791)→  275

Device tag
Navigation

 Expert → Diagnostics → Device info → Device tag (10799)

Description

Use this function to enter a unique name for the measuring point so it can be identified
quickly within the plant. The name is displayed in the header:
1

XXXXXXXXX

A0013375

9

Header text

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /)Factory setting

Prowirl 200

Endress+Hauser

271

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

User entry
The number of characters displayed depends on the characters used.

Serial number
Navigation

 Expert → Diagnostics → Device info → Serial number (10798)

Description

Displays the serial number of the measuring device. It can also be found on the nameplateof the sensor and transmitter.
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Factory settingThis information varies depending on the device. Only an example is given here.

Device type
Navigation

 Expert → Diagnostics → Device info → Device type (10711)

Description

Use this function to view the device type with which the measuring device is registeredwith FOUNDATION Fieldbus.

User interface

Prowirl 200

Device Revision
Navigation

 Expert → Diagnostics → Device info → Device Revision (10710)

Description

Use this function to view the device revision with which the measuring device is registeredwith FOUNDATION Fieldbus.

User interface

1

DD Revision
Navigation

 Expert → Diagnostics → Device info → DD Revision (10709)

Description

Displays the revision number of the device description (DD).

272

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

User interface

1

Additional information

Description

Description of device parameters

This display can be used to ensure that the right system files (DD = device description)
are used for integration into the host system. The system files can be downloaded freeof charge online at: www.endress.com.

Firmware version
Navigation

 Expert → Diagnostics → Device info → Firmware version (10792)

Description

Displays the device firmware version installed.

User interface

Character string with the following format:
xx.yy.zz

Hardware revision
Navigation

 Expert → Diagnostics → Device info → Hardware rev. (10793)

Description

Displays the Hardware revision.

User interface

Character string in the format xx.yy.zz

ITK Version
Navigation

 Expert → Diagnostics → Device info → ITK Version (10794)

Description

Displays the revision status of the Interoperability Test Kits (ITK).

User interface

6

Extended order code
Navigation

 Expert → Diagnostics → Device info → Ext. order cd. (10795)

Description

Use this function to enter the device order code.
The order code can be found on the nameplate of the sensor and transmitter in the"Order code" field.

User entry

Endress+Hauser

Character string composed of letters, numbers and certain punctuation marks

273

Description of device parameters

Additional information

Proline Prowirl 200 FOUNDATION Fieldbus

Description
The order code is generated from the extended order code, which defines all the device
features of the product structure. In contrast, the device features cannot be read directlyfrom the order code.
Uses of the order code
• To order an identical spare device.
• To identify the device quickly and easily, e.g. when contacting Endress+Hauser.

Extended order code 1
Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 1 (10796)

Description

Use this function to enter the first part of the extended order code.
On account of length restrictions, the extended order code is split into a maximum of 2
parameters.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User entry

Character string

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

Extended order code 2
Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 2 (10797)

Description

Use this function to enter the second part of the extended order code.
Deletion of the ordered customer-specific parameter set that was preconfigured
before delivery. This parameter set cannot be reset. It can only be restored to thefactory setting.

User entry

Character string

Additional information

For additional information, see Extended order code 1 parameter

274

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

ENP version
Navigation

 Expert → Diagnostics → Device info → ENP version (10791)

Description

Displays the version of the electronic nameplate. This electronic nameplate stores a data
record for device identification that includes more data than the nameplates attached tothe outside of the device.

User interface

Character string in the format xx.yy.zz

3.10.4"Sensor information" submenu

Navigation

 Expert → Diagnostics → Sensor info

‣ Sensor information
DSC sensor serial number (7728)→  275

DSC sensor serial number
Navigation

 Expert → Diagnostics → Sensor info → DSC serial no. (7728)

Description

Displays the serial number of the DSC sensor that is used in the measuring tube.

User interface

Character string

Additional information

Description
The serial number and other individual values of the DSC sensor, such as temperature
range and reference values, are stored on the S-DAT.
If the DSC sensor is replaced, the S-DAT must also always be replaced.

3.10.5"Data logging" submenu

Navigation

 Expert → Diagnostics → Data logging‣ Data logging

Endress+Hauser

Assign channel 1 (0851)→  276

Assign channel 2 (0852)→  277275

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign channel 3 (0853)→  277

Assign channel 4 (0854)→  278

Logging interval (0856)→  278

Clear logging data (0855)→  279‣ Display channel 1→  279‣ Display channel 2→  280‣ Display channel 3→  281‣ Display channel 4→  281

Assign channel 1Navigation

 Expert → Diagnostics → Data logging → Assign chan. 1 (0851)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Use this function to select a process variable for the data logging channel.

Selection• Off• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *• Reynolds number *• Density *• Vortex frequency• Electronic temperature

Factory setting

Off

*

276

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
A total of 1000 measured values can be logged. This means:• 1000 data points if 1 logging channel is used
• 500 data points if 2 logging channels are used
• 333 data points if 3 logging channels are used
• 250 data points if 4 logging channels are used
Once the maximum number of data points is reached, the oldest data points in the data log
are cyclically overwritten in such a way that the last 1000, 500, 333 or 250 measured
values are always in the log (ring memory principle).
The log contents are cleared if the option selected is changed.

Assign channel 2

Navigation

 Expert → Diagnostics → Data logging → Assign chan. 2 (0852)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Options for the assignment of a process variable to the data logging channel.

Selection

Picklist, see Assign channel 1 parameter (→  276)

Factory setting

Off

Assign channel 3

Navigation

 Expert → Diagnostics → Data logging → Assign chan. 3 (0853)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Options for the assignment of a process variable to the data logging channel.

Selection

Picklist, see Assign channel 1 parameter (→  276)

Factory setting

Off

Endress+Hauser

277

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Assign channel 4Navigation

 Expert → Diagnostics → Data logging → Assign chan. 4 (0854)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Options for the assignment of a process variable to the data logging channel.

Selection

Picklist, see Assign channel 1 parameter (→  276)

Factory setting

Off

Logging interval

Navigation

 Expert → Diagnostics → Data logging → Logging interval (0856)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Use this function to enter the logging interval tlog for data logging.

User entry

1.0 to 3 600.0 s

Factory setting

10.0 sAdditional information

Description
This defines the interval between the individual data points in the data log, and thus the
maximum loggable process time Tlog:• If 1 logging channel is used: Tlog = 1000 × tlog
• If 2 logging channels are used: Tlog = 500 × tlog
• If 3 logging channels are used: Tlog = 333 × tlog
• If 4 logging channels are used: Tlog = 250 × tlog
Once this time elapses, the oldest data points in the data log are cyclically overwritten suchthat a time of Tlog always remains in the memory (ring memory principle).
The log contents are cleared if the length of the logging interval is changed.
ExampleIf 1 logging channel is used:• Tlog = 1000 × 1 s = 1 000 s ≈15 min• Tlog = 1000 × 10 s = 10 000 s ≈ 3 h• Tlog = 1000 × 80 s = 80 000 s ≈ 1 d• Tlog = 1000 × 3 600 s = 3 600 000 s ≈ 41 d

278

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Clear logging dataNavigation

 Expert → Diagnostics → Data logging → Clear logging (0855)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).

Description

Option to clear the entire logging data.

Selection• Cancel• Clear data

Factory setting

CancelAdditional information

Selection• Cancel
The data is not cleared. All the data is retained.
• Clear data
The logging data is cleared. The logging process starts from the beginning.

"Display channel 1" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 1▸ Display channel 1
Display channel 1→  279

Display channel 1
Navigation

Expert → Diagnostics → Data logging → Displ.channel 1

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  54).
One of the following options is selected in the Assign channel 1 parameter (→  276):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature• Calculated saturated steam pressure *• Steam quality **Visibility depends on order options or device settings

Endress+Hauser

279

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *• Reynolds number *• Density *• Pressure *• Vortex frequency• Electronic temperature
Description

Displays the measured value trend for the logging channel in the form of a chart.

Additional information

Description

A0016222• x-axis: depending on the number of channels selected displays 250 to 1000 measured
values of a process variable.
• y-axis: displays the approximate measured value span and constantly adapts this to theongoing measurement.

"Display channel 2" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 2▸ Display channel 2
Display channel 2→  280

Display channel 2
Navigation

Prerequisite

A process variable is defined in the Assign channel 2 parameter.

Description

See the Display channel 1 parameter →  279*

280

Expert → Diagnostics → Data logging → Displ.channel 2

Visibility depends on order options or device settings

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters"Display channel 3" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 3▸ Display channel 3
Display channel 3→  281

Display channel 3
Navigation

Expert → Diagnostics → Data logging → Displ.channel 3

Prerequisite

A process variable is defined in the Assign channel 3 parameter.

Description

See the Display channel 1 parameter →  279"Display channel 4" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 4▸ Display channel 4
Display channel 4→  281

Display channel 4
Navigation

Expert → Diagnostics → Data logging → Displ.channel 4

Prerequisite

A process variable is defined in the Assign channel 4 parameter.

Description

See the Display channel 1 parameter →  279

3.10.6"Min/max values" submenu

Navigation

 Expert → Diagnostics → Min/max val.

‣ Min/max values
Reset min/max values (7706)

Endress+Hauser→  282281

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus‣ Terminal voltage→  282‣ IO module temperature→  283‣ Pre-amplifier temperature→  285‣ Medium temperature→  286‣ Flow velocity→  286‣ External pressure→  287

Reset min/max values



Navigation

 Expert → Diagnostics → Min/max val. → Reset min/max (7706)

Description

Use this function to select measured variables whose minimum, maximum and averagemeasured values are to be reset.

Selection• Cancel• Terminal voltage 1• Temperature• Flow velocity• Pressure

Factory setting

Cancel"Terminal voltage" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Terminal volt.

‣ Terminal voltage

282

Minimum value (0689)→  283

Maximum value (0663)→  283

Average value (0698)→  283

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Terminal volt. → Minimum value (0689)

Description

Use this function to display the smallest previously measured terminal voltage value in
Volts.

User interface

0.0 to 50.0 V

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Terminal volt. → Maximum value (0663)

Description

Use this function to view the largest previously measured terminal voltage value in Volts.

User interface

0.0 to 50.0 V

Average value
Navigation

 Expert → Diagnostics → Min/max val. → Terminal volt. → Average value (0698)

Description

Use this function to view the average of all previously measured terminal voltage values in
Volts.

User interface

Signed floating-point number"IO module temperature" submenu
Navigation

 Expert → Diagnostics → Min/max val. → IO module temp.

‣ IO module temperature

Endress+Hauser

Minimum value (0688)→  284

Maximum value (0665)→  284

Average value (0697)→  284283

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → IO module temp. → Minimum value (0688)

Description

Displays the lowest previously measured temperature value of the I/O electronics module.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75)

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → IO module temp. → Maximum value (0665)

Description

Displays the highest previously measured temperature value of the I/O electronics module.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75)

Average value
Navigation

 Expert → Diagnostics → Min/max val. → IO module temp. → Average value (0697)

Description

Displays the average value of all previously measured temperature values of the I/O
electronics module.

User interface–1 273.15 to 726.85 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75)

284

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters"Pre-amplifier temperature" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Pre-amplif. temp

‣ Pre-amplifier temperature
Minimum value (7724)→  285

Maximum value (7723)→  285

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Pre-amplif. temp → Minimum value (7724)

Description

Displays the lowest previously measured temperature value of the pre-amplifier module.

User interface

0 to 1 000 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75).

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Pre-amplif. temp → Maximum value(7723)

Description

Displays the highest previously measured temperature value of the pre-amplifier module.

User interface

0 to 1 000 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75).

Endress+Hauser

285

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus"Medium temperature" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp.

‣ Medium temperature
Minimum value (7655)→  286

Maximum value (7654)→  286

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp. → Minimum value (7655)

Description

Displays the lowest previously medium temperature.

User interface

0 to 1 000 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75).

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp. → Maximum value (7654)

Description

Displays the highest previously medium temperature.

User interface

0 to 1 000 °C

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  75).

"Flow velocity" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Flow velocity‣ Flow velocity
Maximum value (7633)

286→  287

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Flow velocity → Maximum value (7633)

Description

Displays the highest previously measured flow velocity.

User interface

Positive floating-point number

Additional information

Dependency
The unit is taken from the Velocity unit parameter (→  79)"External pressure" submenu
Navigation

 Expert → Diagnostics → Min/max val. → External press.

‣ External pressure
Maximum value (7623)→  287

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → External press. → Maximum value (7623)

Description

Displays the highest previously measured external pressure.

User interface

Positive floating-point number

Additional information

Dependency
The unit is taken from the Pressure unit parameter (→  74)

3.10.7"Heartbeat" submenu

For detailed information on the parameter descriptions of the Heartbeat Verification
application package, see the Special Documentation for the device
Navigation

 Expert → Diagnostics → Heartbeat‣ Heartbeat‣ Heartbeat base settings
Endress+Hauser

287

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus‣ Performing verification‣ Verification results

3.10.8"Simulation" submenu

Navigation

 Expert → Diagnostics → Simulation‣ Simulation
Assign simulation process variable(1810)→  288

Value process variable (1811)→  289

Frequency simulation (0472)→  290

Frequency value (0473)→  290

Pulse simulation (0458)→  290

Pulse value (0459)→  291

Switch output simulation (0462)→  291

Switch status (0463)→  292

Simulation device alarm (0654)→  292

Diagnostic event category (0738)→  293

Simulation diagnostic event (0737)→  293

Assign simulation process variable

Navigation

 Expert → Diagnostics → Simulation → Assign proc.var. (1810)

Description

Use this function to select a process variable for the simulation process that is activated.
The display alternates between the measured value and a diagnostic message of the"Function check" category (C) while simulation is in progress.

Selection• Off• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature

288

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Condensate mass flow *• Energy flow
• Heat flow difference *• Reynolds number
Factory setting

Off

Additional information

Description
The simulation value of the selected process variable is specified in the Value process
variable parameter (→  289).

Value process variable

Navigation

 Expert → Diagnostics → Simulation → Value proc. var. (1811)

Prerequisite

One of the following options is selected in the Assign simulation process variable
parameter (→  288):• Volume flow
• Corrected volume flow
• Mass flow
• Flow velocity• Temperature *• Calculated saturated steam pressure *• Steam quality *• Total mass flow *• Condensate mass flow *• Energy flow *• Heat flow difference *• Reynolds number *Description

Use this function to enter a simulation value for the selected process variable. Subsequent
measured value processing and the signal output use this simulation value. In this way,users can verify whether the measuring device has been configured correctly.

User entry

Depends on the process variable selected

Factory setting0

Additional information

User entry
The unit of the displayed measured value is taken from the System units submenu(→  69).

*

Visibility depends on order options or device settings

Endress+Hauser

289

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Frequency simulation



Navigation

 Expert → Diagnostics → Simulation → Frequency sim. (0472)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  121).

Description

Use this function to switch simulation of the frequency output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Frequency value parameter(→  290).
Selection• Off
Frequency simulation is switched off. The device is in normal measuring mode oranother process variable is being simulated.
• On
Frequency simulation is active.

Frequency value

Navigation

 Expert → Diagnostics → Simulation → Freq. value (0473)

Prerequisite

The On option is selected in the Frequency simulation parameter (→  290).

Description

Use this function to enter a frequency value for the simulation. In this way, users can verify
the correct adjustment of the frequency output and the correct function of downstream
switching units.

User entry

0.0 to 1 250.0 Hz

Factory setting

0.0 HzPulse simulation



Navigation

 Expert → Diagnostics → Simulation → Pulse sim. (0458)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  121).

290

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to switch simulation of the pulse output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection• Off• Fixed value• Down-counting value

Factory setting

Off

Additional information

Description
The desired simulation value is specified in the Pulse value parameter (→  291).
Selection• Off
Pulse simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Fixed value
Pulses with the pulse width specified in the Pulse width parameter (→  124) are
output continuously.
• Down-counting value
The pulses specified in the Pulse value parameter (→  291) are output.

Pulse value

Navigation

 Expert → Diagnostics → Simulation → Pulse value (0459)

Prerequisite

In the Pulse simulation parameter (→  290), the Down-counting value option is
selected.

Description

Use this function to enter a pulse value for the simulation. In this way, users can verify thecorrect adjustment of the pulse output and the correct function of downstream switchingunits.

User entry

0 to 65 535

Switch output simulation



Navigation

 Expert → Diagnostics → Simulation → Switch sim. (0462)

Prerequisite

The Switch option is selected in the Operating mode parameter (→  121).

Description

Use this function to switch simulation of the switch output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Endress+Hauser

291

Description of device parameters

Proline Prowirl 200 FOUNDATION Fieldbus

Factory setting

Off

Additional information

Description
The desired simulation value is specified in the Switch status parameter (→  292).
Selection• Off
Switch simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Switch simulation is active.

Switch status



Navigation

 Expert → Diagnostics → Simulation → Switch status (0463)

Prerequisite

The On option is selected in the Switch output simulation parameter (→  291).

Description

Use this function to select a switch value for the simulation. In this way, users can verify
the correct adjustment of the switch output and the correct function of downstream
switching units.

Selection• Open• Closed

Factory setting

Open

Additional information

Options• OpenSwitch simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Closed
Switch simulation is active.

Simulation device alarmNavigation

 Expert → Diagnostics → Simulation → Sim. alarm (0654)

Description

Use this function to switch the device alarm on and off.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
In this way, users can verify the correct function of downstream switching units.

292

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Description of device parameters

The display alternates between the measured value and a diagnostic message of the"Function check" category (C) while simulation is in progress.

Diagnostic event category
Navigation

Expert → Diagnostics → Simulation → Event category (0738)

Description

Use this function to select the category of the diagnostic events that are displayed for thesimulation in the Simulation diagnostic event parameter (→  293).

Selection• Sensor• Electronics• Configuration• ProcessFactory setting

Sensor

Simulation diagnostic event
Navigation

 Expert → Diagnostics → Simulation → Sim. diag. event (0737)

Description

Use this function to select a diagnostic event for the simulation process that is activated.

Selection• Off• Diagnostic event picklist (depends on the category selected)

Factory setting

Off

Additional information

Description
For the simulation, you can choose from the diagnostic events of the category selectedin the Diagnostic event category parameter (→  293).

Endress+Hauser

293

Country-specific factory settings

Proline Prowirl 200 FOUNDATION Fieldbus

4Country-specific factory settings

4.1

SI units
Not valid for USA and Canada.

4.1.1

System units

Volume flowm³/hVolumem³

Mass flow

kg/h

Mass

kg

Corrected volume flow

Nm³/hCorrected volume

Nm³

Pressurebar

Temperature°C

Energy flow

kW

EnergykWhCalorific value (volume)

kJ/Nm³

Calorific value (mass)

kJ/kg

Velocitym/s

Densitykg/m³

Dynamic viscosity

Pa sSpecific heat capacity

kJ/(kgK)

Lengthmm

4.1.2

Full scale values

The factory settings applie to the following parameters:
100% bar graph value 1
Nominal diameter
[mm]

294

[m3/h]

15
25 > 1540 >> 1525

25
40 > 25
50 >> 25125

40
50 > 40
80 >> 40308

5080 > 50
100 >> 50

513

80
100 > 80
150 >> 80

1 152

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Country-specific factory settings

Nominal diameter[mm]
100
150 > 100
200 >> 100

1 995

150
200 > 150
250 >> 150

4 539

200
250 > 200
300 >> 200

8 713

250
300 > 250
350 >> 250

13 735300
350 > 300
400 >> 300

19 701

4.1.3

Endress+Hauser[m3/h]

Pulse value

NominaldiameterVolume flow

Mass flow[mm]( 2 pulse/s)[m³/pulse]( 2 pulse/s)[kg/pulse]

15
25 > 1540 >> 150.00067

0.003425
40 > 25
50 >> 25

0.00350.018

4050 > 40
80 >> 400.00850.044

5080 > 50
100 >> 50

0.023

0.12

80
100 > 80
150 >> 80

0.051

0.26

100
150 > 100
200 >> 100

0.089

0.46

150
200 > 150
250 >> 150

0.20

1.04

200
250 > 200
300 >> 200

0.39

1.99

250
300 > 250
350 >> 250

0.61

3.14

300
350 > 300
400 >> 300

0.88

4.51

295

Country-specific factory settings

Proline Prowirl 200 FOUNDATION Fieldbus

4.2

US units
Only valid for USA and Canada.

4.2.1

System units

Volume flow

ft³/minVolumeft³

Mass flow

lb/minMasslb

Corrected volume flow

Sft³/minCorrected volume

Sft³

Pressure

psi

Temperature°F

Energy flow

Btu/hEnergyBtu

Calorific value (volume)

Btu/Sft³

Calorific value (mass)

Btu/lb

Velocity

ft/s

Densitylb/ft³

Lengthin

4.2.2

Full scale values

The factory settings applie to the following parameters:
100% bar graph value 1
Nominal diameter
[in]

296

[ft3/h]½1>½
1½ >> ½882

1
1½ > 1
2 >> 14 414

1½
2 > 1½
3 >> 1½

10 8762
3>2
4 >> 218 1163
4>3
6 >> 340 682

46>48 >> 470 452

6
8>6
10 >> 6160 293

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Country-specific factory settings

Nominal diameter[in]810 > 812 >> 8307 6961012 > 10
14 >> 10485 04612
14 > 12
16 >> 12695 734

4.2.3

Endress+Hauser[ft3/h]Pulse value

NominaldiameterVolume flow

Volume flow[in]

 2 pulse/s[gal/pulse] 2 pulse/s[lb/pulse]½
1>½
1½ >> ½

0.18

0.00761
1½ > 1
2 >> 1

0.92

0.039

1½
2 > 1½
3 >> 1½

2.25

0.097

2
3>2
4 >> 2

6.02

0.26

3
4>3
6 >> 3

13.50

0.58

4
6>4
8 >> 4

23.42

1.01

6
8>6
10 >> 6

53.29

2.29

8
10 > 812 >> 8102.294.40

1012 > 10
14 >> 10161.26

6.93

1214 > 12
16 >> 12231.309.94

297

Explanation of abbreviated units

5

Explanation of abbreviated units

5.1

SI units

Process variable Units

Explanation

Calorific value(volume)

kWh/Nm³, MWh/Nm³, kJ/Nm³,
MJ/Nm³

Kilowatt hour, megawatt hour, kilojoule, megajoule/standard cubic meter

kWh/Sm³, MWh/Sm³, kJ/Sm³,
MJ/Sm³

Kilowatt hour, megawatt hour, kilojoule, megajoule/standard cubic meter

Calorific value(mass)

kWh/kg, MWh/kg, kJ/kg, MJ/kg

Kilowatt hour, megawatt hour, kilojoule, megajoule/kilogramDensityg/cm³

Gram/volume unit

kg/dm³, kg/l, kg/m³

Kilogram/volume unit

SD4°C, SD15°C, SD20°C

Specific density: The specific density is the ratio of thefluid density to the water density at a watertemperature of 4 °C (39 °F), 15 °C (59 °F), 20 °C (68 °F).

SG4°C, SG15°C, SG20°C

Specific gravity: The specific gravity is the ratio of thefluid density to the water density at a watertemperature of 4 °C (39 °F), 15 °C (59 °F), 20 °C (68 °F).

Pa, kPa, MPa

Pascal, kilopascal, megapascal

mbar aMillibar (absolute)bar, torr, atm

Bar, torr, physical atmosphere

gf/cm², kgf/cm²

Gram force, kilogram force/square centimeter

Dynamic
viscosity

Pa s

Pascal second

cP, P

Centipoise, poise

EnergykWh, MWh, GWh

Kilowatt hour, megawatt hour, gigawatt hourkJ, MJ, GJ

Kilojoule, megajoule, gigajoulekcal, Mcal, Gcal

Kilocalories, megacalories, gigacalories

kW, MW, GW

Kilowatt, megawatt

kJ/s, kJ/min, kJ/h, kJ/d

Kilojoule/time unit

MJ/s, MJ/min, MJ/h, MJ/d

Megajoule/time unit

GJ/s, GJ/min, GJ/h, GJ/d

Gigajoule/time unit

kcal/s, kcal/min, kcal/h, kcal/d

Kilocalories/time unit

Mcal/s, Mcal/min, Mcal/h, Mcal/d

Megacalories/time unit

Gcal/s, Gcal/min, Gcal/h, Gcal/d

Gigacalories/time unit

Velocitym/s

Meter/time unit

Lengthmm, mMillimeter, meter

Massg, kg, tGram, kilogram, metric ton

Mass flow

g/s, g/min, g/h, g/dGram/time unit

kg/s, kg/min, kg/h, kg/d

Kilogram/time unit

t/s, t/min, t/h, t/dMetric ton/time unit

Corrected
volumeNl, Nm³, Sm³

Normal liter, normal cubic meter, standard cubic meter

Corrected
volume flow

Nl/s, Nl/min, Nl/h, Nl/d

Normal liter/time unit

Nm³/s, Nm³/min, Nm³/h, Nm³/d

Normal cubic meter/time unit

Sm³/s, Sm³/min, Sm³/h, Sm³/d

Standard cubic meter/time unit

Pressure

Energy flow

298

Proline Prowirl 200 FOUNDATION Fieldbus

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Process variable Units

Explanation

Specific heat
capacity

kJ/(kgK), MJ/(kgK)

Kilojoule, megajoule/kilogram KelvinkWh/(kgK)

Kilowatt hour/kilogram Kelvinkcal/(kgK)

Kilocalories/kilogram Kelvin

Temperature°C , K

Celsius, Kelvin

Volumecm³, dm³, m³

Cubic centimeter, cubic decimeter, cubic meterml, lMilliliter, liter

cm³/s , cm³/min, cm³/h, cm³/d

Cubic centimeter/time unit

dm³/s, dm³/min, dm³/h, dm³/dCubic decimeter/time unit

m³/s, m³/min, m³/h, m³/d

Cubic meter/time unit

ml/s, ml/min, ml/h, ml/dMilliliter/time unit

l/s, l/min, l/h, l/d

Liter/time unitTimem, h, d, y

Minute, hour, day, year

5.2

US units

Volume flow

Process variable Units

Explanation

Calorific value(mass)

kWh/lb, MWh/lb, kJ/lb, MJ/lb

Kilowatt hour, kilojoule, British thermal unit, thousand
British thermal units/pound

Densitylb/ft³, lb/gal (us)Pound/cubic foot, pound/gallon

lb/bbl (us;liq.), lb/bbl (us;beer),lb/bbl (us;oil), lb/bbl (us;tank)

Pound/volume unit

Pressure

psi a

Psi absolute

Velocity

ft/s

Foot/time unit

Lengthin, ftInch, footMassoz, lb, STon

Ounce, pound, standard ton

Mass flow

oz/s, oz/min, oz/h, oz/dOunce/time unit

lb/s, lb/min, lb/h, lb/dPound/time unit

STon/s, STon/min, STon/h, STon/dStandard ton/time unit

Corrected
volumeSft³Standard cubic footCorrected
volume flow

Sft³/s, Sft³/min, Sft³/h, Sft³/dStandard cubic foot/time unit

Temperature°F, °R

Fahrenheit, Rankine

Volumeaf

Acre footft³

Cubic footfl oz (us), gal (us), kgal (us), Mgal
(us)

Fluid ounce, gallon, kilogallon, million gallonbbl (us;liq.), bbl (us;beer), bbl(us;oil), bbl (us;tank)Barrel (normal liquids), barrel (beer), barrel(petrochemicals), barrel (filling tanks)

af/s, af/min, af/h, af/dAcre foot/time unit

ft³/s, ft³/min, ft³/h, ft³/d

Cubic foot/time unit

fl oz/s (us), fl oz/min (us), fl oz/h(us), fl oz/d (us)Fluid ounce/time unit

gal/s (us), gal/min (us), gal/h (us,)
gal/d (us)Gallon/time unit

Volume flow

Endress+Hauser

Explanation of abbreviated units

299

Explanation of abbreviated units

Proline Prowirl 200 FOUNDATION Fieldbus

Process variable Units

Time

5.3

Explanation

kgal/s (us), kgal/min (us), kgal/h(us,) kgal/d (us)Kilogallon/time unit

Mgal/s (us), Mgal/min (us),Mgal/h (us), Mgal/d (us)Million gallon/time unit

bbl/s (us;liq.), bbl/min (us;liq.),bbl/h (us;liq.), bbl/d (us;liq.)

Barrel/time unit (normal liquids)
Normal liquids: 31.5 gal/bbl

bbl/s (us;beer), bbl/min (us;beer),bbl/h (us;beer), bbl/d (us;beer)

Barrel /time unit (beer)
Beer: 31.0 gal/bbl

bbl/s (us;oil), bbl/min (us;oil),bbl/h (us;oil), bbl/d (us;oil)Barrel/time unit (petrochemicals)Petrochemicals: 42.0 gal/bbl

bbl/s (us;tank), bbl/min (us;tank),
bbl/h (us;tank), bbl/d (us;tank)Barrel/time unit (filling tank)
Filling tanks: 55.0 gal/bbl

m, h, d, y

Minute, hour, day, year

am, pmAnte meridiem ( before midday), post meridiem (after
midday)Imperial units

Process variable Units

Explanation

Calorific value(volume)

Btu/Sm³, MBtu/Sm³

British thermal unit, thousand British thermal units/
standard cubic meter

Btu/Sft³, MBtu/Sft³

British thermal unit, thousand British thermal units/
standard cubic footCalorific value(mass)

Btu/lb, MBtu/lb

British thermal unit, thousand British thermal units/
pound

Densitylb/gal (imp), lb/bbl (imp;beer),lb/bbl (imp;oil)Pound/volume unit

EnergyBtu, MBtu, MMBtu

British thermal unit, thousand British thermal units,
million British thermal units

Energy flow

Btu/s, Btu/min, Btu/h, Btu/day

British thermal unit/time unit

MBtu/s, MBtu/min, MBtu/h,
MBtu/dThousand British thermal units/time unit

MMBtu/s, MMBtu/min, MMBtu/h, Million British thermal units/time unit
MMBtu/dSpecific heat
capacity

Btu/(lb °R)

British thermal unit/pound degree Rankine

Volumegal (imp), Mgal (imp)

Gallon, mega gallonbbl (imp;beer), bbl (imp;oil)Barrel (beer), barrel (petrochemicals)

gal/s (imp), gal/min (imp), gal/h(imp), gal/d (imp)

Gallon/time unit

Mgal/s (imp), Mgal/min (imp),
Mgal/h (imp), Mgal/d (imp)

Mega gallon/time unit

bbl/s (imp;beer), bbl/min(imp;beer), bbl/h (imp;beer), bbl/d(imp;beer)

Barrel /time unit (beer)
Beer: 36.0 gal/bbl

bbl/s (imp;oil), bbl/min (imp;oil),bbl/h (imp;oil), bbl/d (imp;oil)Barrel/time unit (petrochemicals)Petrochemicals: 34.97 gal/bbl

m, h, d, y

Minute, hour, day, year

am, pmAnte meridiem ( before midday), post meridiem (after
midday)Volume flow

Time

300

Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

5.4

Other units

Process variable Units

Explanation

Pressure

mmH2O (4°C)

Millimeter of water column (4 °C)

mmH2O (68°F)

Millimeter of water column (68 °F)

mmHg (0°C)

Millimeter of mercury column (0 °C)

inH2O (4°C)

Inch of water column (4 °C)

inH2O (68°F)

Inch of water column (68 °F)

ftH2O (68°F)

Foot of water column (68 °F)

inHg (0°C)

Inch of mercury (0 °C)

m³/kg

Cubic meter/kilogram

ft³/lb

Cubic foot/pound

Specific volume

Endress+Hauser

Explanation of abbreviated units

301

Index

Proline Prowirl 200 FOUNDATION Fieldbus

Index
0…9

0% bargraph value 1 (Parameter) . . . . . . . . . . . . . . . 18
0% bargraph value 3 (Parameter) . . . . . . . . . . . . . . . 21
2nd temperature delta heat (Parameter) . . . . . . . . . 115
100% bargraph value 1 (Parameter) . . . . . . . . . . . . . 19
100% bargraph value 3 (Parameter) . . . . . . . . . . . . . 22

A

Access status display (Parameter) . . . . . . . . . . . . 12, 26
Access status tooling (Parameter) . . . . . . . . . . . . . . . 13
Acknowledge Option (Parameter) . . . . . . 158, 200, 233
Activate sensor emergency mode (Parameter) . . . . . . 54
Activate SW option (Parameter) . . . . . . . . . . . . . . . . 53
Actual diagnostics (Parameter) . . . . . . . . . . . . . . . . 262
Actual mode (Parameter) . . . . . 143, 163, 213, 241, 250
Administration (Submenu) . . . . . . . . . . . . . . . . . . . . 50
Alarm delay (Parameter) . . . . . . . . . . . . . . . . . . . . . . 31
Alarm Hysteresis (Parameter) . . . . . . . . . . . . . . . . . 201
Alarm State (Parameter) 155, 159, 198, 204, 206,
207,
209,
235,
236
Alert Key (Parameter) . . . . . . . 141, 163, 212, 240, 249
Analog input 1 to 4 (Submenu) . . . . . . . . . . . . . . . . 161
Analog inputs (Submenu) . . . . . . . . . . . . . . . . . . . . 161
Analog outputs (Submenu) . . . . . . . . . . . . . . . . . . . 239
Application (Submenu) . . . . . . . . . . . . . . . . . . . . . . 256
Assign behavior of diagnostic no. 022 (Parameter) . . . 34
Assign behavior of diagnostic no. 122 (Parameter) . . . 35
Assign behavior of diagnostic no. 350 (Parameter) . . . 35
Assign behavior of diagnostic no. 371 (Parameter) . . . 36
Assign behavior of diagnostic no. 442 (Parameter) . . . 36
Assign behavior of diagnostic no. 443 (Parameter) . . . 36
Assign behavior of diagnostic no. 828 (Parameter) . . . 37
Assign behavior of diagnostic no. 829 (Parameter) . . . 37
Assign behavior of diagnostic no. 832 (Parameter) . . . 37
Assign behavior of diagnostic no. 833 (Parameter) . . . 38
Assign behavior of diagnostic no. 834 (Parameter) . . . 38
Assign behavior of diagnostic no. 835 (Parameter) . . . 39
Assign behavior of diagnostic no. 841 (Parameter) . . . 39
Assign behavior of diagnostic no. 844 (Parameter) . . . 39
Assign behavior of diagnostic no. 870 (Parameter) . . . 40
Assign behavior of diagnostic no. 871 (Parameter) . . . 40
Assign behavior of diagnostic no. 872 (Parameter) . . . 40
Assign behavior of diagnostic no. 873 (Parameter) . . . 41
Assign behavior of diagnostic no. 945 (Parameter) . . . 41
Assign behavior of diagnostic no. 947 (Parameter) . . . 42
Assign channel 1 (Parameter) . . . . . . . . . . . . . . . . . 276
Assign channel 2 (Parameter) . . . . . . . . . . . . . . . . . 277
Assign channel 3 (Parameter) . . . . . . . . . . . . . . . . . 277
Assign channel 4 (Parameter) . . . . . . . . . . . . . . . . . 278
Assign diagnostic behavior (Parameter) . . . . . . . . . . 133
Assign flow direction check (Parameter) . . . . . . . . . 136
Assign frequency output (Parameter) . . . . . . . . . . . . 127
Assign limit (Parameter) . . . . . . . . . . . . . . . . . . . . . 134
Assign process variable (Parameter) . . . . . . . . . . 83, 257
Assign pulse output (Parameter) . . . . . . . . . . . . . . . 123
Assign simulation process variable (Parameter) . . . . 288
302

Assign status (Parameter) . . . . . . . . . . . . . . . . . . . . 137
Assign status of diagnostic number 022 (Parameter) . 42
Assign status of diagnostic number 122 (Parameter) . 42
Assign status of diagnostic number 350 (Parameter) . 43
Assign status of diagnostic number 371 (Parameter) . 43
Assign status of diagnostic number 828 (Parameter) . 43
Assign status of diagnostic number 829 (Parameter) . 44
Assign status of diagnostic number 832 (Parameter) . 44
Assign status of diagnostic number 833 (Parameter) . 44
Assign status of diagnostic number 834 (Parameter) . 45
Assign status of diagnostic number 835 (Parameter) . 45
Assign status of diagnostic number 841 (Parameter) . 45
Assign status of diagnostic number 844 (Parameter) . 46
Assign status of diagnostic number 870 (Parameter) . 46
Assign status of diagnostic number 871 (Parameter) . 46
Assign status of diagnostic number 872 (Parameter) . 47
Assign status of diagnostic number 873 (Parameter) . 47
Assign status of diagnostic number 945 (Parameter) . 47
Assign status of diagnostic number 947 (Parameter) . 48
Atmospheric pressure (Parameter) . . . . . . . . . . . . . 113
Average value (Parameter) . . . . . . . . . . . . . . . 283, 284

BBacklight (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 26
Backup state (Parameter) . . . . . . . . . . . . . . . . . . . . . 29
Block Error (Parameter) . . . . . . 144, 165, 214, 243, 252
Block tag (Parameter) . . . . . . . 140, 162, 211, 239, 248CCalculated saturated steam pressure (Parameter) . . . . 60
Calibration (Submenu) . . . . . . . . . . . . . . . . . . . . . . 119
Calibration factor (Parameter) . . . . . . . . . . . . . . . . . 119
Calorific value (Parameter) . . . . . . . . . . . . . . . . . . . . 92
Calorific value type (Parameter) . . . . . . . . . . . . . . . . . 90
Calorific value unit (Parameter) . . . . . . . . . . . . . . 77, 78
Channel (Parameter) . . . . . . . . . . . . 190, 228, 243, 252
Clear Fault State (Parameter) . . . . . . . . . . . . . . . . . . 151
Clear logging data (Parameter) . . . . . . . . . . . . . . . . 279
Communication (Submenu) . . . . . . . . . . . . . . . . . . . 139
Comparison result (Parameter) . . . . . . . . . . . . . . . . . 29
Compressibility factor (Parameter) . . . . . . . . . . . . . . . 64
Condensate mass flow (Parameter) . . . . . . . . . . . . . . 61
Configuration backup display (Submenu) . . . . . . . . . . 27
Configuration management (Parameter) . . . . . . . . . . 28
Confirm access code (Parameter) . . . . . . . . . . . . . . . . 51
Confirm Time (Parameter) . . . . . . . . . . . . . . . . . . . . 151
Contrast display (Parameter) . . . . . . . . . . . . . . . . . . . 26
Control Totalizer 1 to 3 (Parameter) . . . . . . . . . . . . 260
Corrected volume flow (Parameter) . . . . . . . . . . . . . . 57
Corrected volume flow unit (Parameter) . . . . . . . . . . . 73
Corrected volume unit (Parameter) . . . . . . . . . . . . . . 74
Current (Parameter) . . . . . . . . . . . . . . . . 157, 199, 233
Cycle Selection (Parameter) . . . . . . . . . . . . . . . . . . . 149
Cycle Type (Parameter) . . . . . . . . . . . . . . . . . . . . . . 149

D

Damping output (Parameter) . . . . . . . . . . . . . . . . . . 130
Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Data logging (Submenu) . . . . . . . . . . . . . . . . . . . . . 275
Date/time format (Parameter) . . . . . . . . . . . . . . . . . . 81
DD Resource (Parameter) . . . . . . . . . . . . . . . . . . . . . 146
DD Revision (Parameter) . . . . . . . . . . . . . . . . . 147, 272
Decimal (Parameter) . . . . . . . . . . . . . . . . . . . . 179, 189
Decimal places 1 (Parameter) . . . . . . . . . . . . . . . . . . 19
Decimal places 2 (Parameter) . . . . . . . . . . . . . . . . . . 20
Decimal places 3 (Parameter) . . . . . . . . . . . . . . . . . . 22
Decimal places 4 (Parameter) . . . . . . . . . . . . . . . . . . 23
Define access code (Parameter) . . . . . . . . . . . . . . 50, 51
Define access code (Wizard) . . . . . . . . . . . . . . . . . . . 50
Degrees of superheat limit (Parameter) . . . . . . . . . . . 49
Delta heat calculation (Parameter) . . . . . . . . . . . . . . 114
Density (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 62
Density calculation (Parameter) . . . . . . . . . . . . . . . . . 88
Density unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 79
Deny (Parameter) . . . . . . . . . . . . . . . . . . 147, 189, 227
Device address (Parameter) . . . . . . . . . . . . . . . . . . . 139
Device information (Submenu) . . . . . . . . . . . . . . . . 271
Device Revision (Parameter) . . . . . . . . . . . . . . 147, 272
Device tag (Parameter) . . . . . . . . . . . . . . . . . . . . . . 271
Device type (Parameter) . . . . . . . . . . . . . . . . . 146, 272
Diagnostic behavior (Submenu) . . . . . . . . . . . . . . . . . 32
Diagnostic event category (Parameter) . . . . . . . . . . . 293
Diagnostic handling (Submenu) . . . . . . . . . . . . . . . . . 30
Diagnostic limits (Submenu) . . . . . . . . . . . . . . . . . . . 48
Diagnostic list (Submenu) . . . . . . . . . . . . . . . . . . . . 265
Diagnostics (Submenu) . . . . . . . . . . . . . . . . . . . . . . 262
Diagnostics 1 (Parameter) . . . . . . . . . . . . . . . . . . . . 265
Diagnostics 2 (Parameter) . . . . . . . . . . . . . . . . . . . . 266
Diagnostics 3 (Parameter) . . . . . . . . . . . . . . . . . . . . 266
Diagnostics 4 (Parameter) . . . . . . . . . . . . . . . . . . . . 267
Diagnostics 5 (Parameter) . . . . . . . . . . . . . . . . . . . . 268
Direct access
0% bargraph value 1 (0123) . . . . . . . . . . . . . . . . . 18
0% bargraph value 3 (0124) . . . . . . . . . . . . . . . . . 21
2nd temperature delta heat (7625) . . . . . . . . . . . 115
100% bargraph value 1 (0125) . . . . . . . . . . . . . . . 19
100% bargraph value 3 (0126) . . . . . . . . . . . . . . . 22
Access status display (0091) . . . . . . . . . . . . . . 12, 26
Access status tooling (0005) . . . . . . . . . . . . . . . . . 13
Acknowledge Option
Analog input 1 to 4 (6910–1 to 4) . . . . . . . . . 200
Discrete input 1 to 2 (6841–1 to 2) . . . . . . . . 233
Acknowledge Option (10691) . . . . . . . . . . . . . . . 158
Activate sensor emergency mode (7712) . . . . . . . . 54
Activate SW option (0029) . . . . . . . . . . . . . . . . . . 53
Actual diagnostics (0691) . . . . . . . . . . . . . . . . . . 262
Actual mode
Analog input 1 to 4 (6957–1 to 4) . . . . . . . . . 163
Discrete input 1 to 2 (6870–1 to 2) . . . . . . . . 213
Actual mode (10725) . . . . . . . . . . . . . . . . . . . . . 143
Actual mode (11264) . . . . . . . . . . . . . . . . . . . . . 250
Actual mode (11366) . . . . . . . . . . . . . . . . . . . . . 241
Alarm delay (0651) . . . . . . . . . . . . . . . . . . . . . . . 31
Alarm Hysteresis
Analog input 1 to 4 (6911–1 to 4) . . . . . . . . . 201
Alarm State
Analog input 1 to 4 (6917–1 to 4) . . . . . . . . . 198
Endress+Hauser

Index

Analog input 1 to 4 (6927–1 to 4) . . . . . . . . . 206
Analog input 1 to 4 (6932–1 to 4) . . . . . . . . . 204
Analog input 1 to 4 (6942–1 to 4) . . . . . . . . . 207
Analog input 1 to 4 (6948–1 to 4) . . . . . . . . . 209
Discrete input 1 to 2 (6847–1 to 2) . . . . . . . . 235
Discrete input 1 to 2 (6858–1 to 2) . . . . . . . . 236
Alarm State (10697) . . . . . . . . . . . . . . . . . . . . . 155
Alarm State (10742) . . . . . . . . . . . . . . . . . . . . . 159
Alert Key
Analog input 1 to 4 (6916–1 to 4) . . . . . . . . . 163
Discrete input 1 to 2 (6846–1 to 2) . . . . . . . . 212
Alert Key (10696) . . . . . . . . . . . . . . . . . . . . . . . 141
Alert Key (11251) . . . . . . . . . . . . . . . . . . . . . . . 249
Alert Key (11365) . . . . . . . . . . . . . . . . . . . . . . . 240
Assign behavior of diagnostic no. 022 (0751) . . . . 34
Assign behavior of diagnostic no. 122 (0752) . . . . 35
Assign behavior of diagnostic no. 350 (0756) . . . . 35
Assign behavior of diagnostic no. 371 (0757) . . . . 36
Assign behavior of diagnostic no. 442 (0658) . . . . 36
Assign behavior of diagnostic no. 443 (0659) . . . . 36
Assign behavior of diagnostic no. 828 (0755) . . . . 37
Assign behavior of diagnostic no. 829 (0754) . . . . 37
Assign behavior of diagnostic no. 832 (0675) . . . . 37
Assign behavior of diagnostic no. 833 (0676) . . . . 38
Assign behavior of diagnostic no. 834 (0677) . . . . 38
Assign behavior of diagnostic no. 835 (0678) . . . . 39
Assign behavior of diagnostic no. 841 (0729) . . . . 39
Assign behavior of diagnostic no. 844 (0747) . . . . 39
Assign behavior of diagnostic no. 870 (0726) . . . . 40
Assign behavior of diagnostic no. 871 (0748) . . . . 40
Assign behavior of diagnostic no. 872 (0746) . . . . 40
Assign behavior of diagnostic no. 873 (0749) . . . . 41
Assign behavior of diagnostic no. 945 (0750) . . . . 41
Assign behavior of diagnostic no. 947 (0753) . . . . 42
Assign channel 1 (0851) . . . . . . . . . . . . . . . . . . 276
Assign channel 2 (0852) . . . . . . . . . . . . . . . . . . 277
Assign channel 3 (0853) . . . . . . . . . . . . . . . . . . 277
Assign channel 4 (0854) . . . . . . . . . . . . . . . . . . 278
Assign diagnostic behavior (0482) . . . . . . . . . . . 133
Assign flow direction check (0484) . . . . . . . . . . . 136
Assign frequency output (0478) . . . . . . . . . . . . . 127
Assign limit (0483) . . . . . . . . . . . . . . . . . . . . . . 134
Assign process variable
Totalizer 1 to 3 (0914–1 to 3) . . . . . . . . . . . . 257
Assign process variable (1837) . . . . . . . . . . . . . . . 83
Assign pulse output (0460) . . . . . . . . . . . . . . . . 123
Assign simulation process variable (1810) . . . . . 288
Assign status (0485) . . . . . . . . . . . . . . . . . . . . . 137
Assign status of diagnostic number 022 (11041) . 42
Assign status of diagnostic number 122 (11042) . 42
Assign status of diagnostic number 350 (11000) . 43
Assign status of diagnostic number 371 (11014) . 43
Assign status of diagnostic number 828 (11015) . 43
Assign status of diagnostic number 829 (11001) . 44
Assign status of diagnostic number 832 (11002) . 44
Assign status of diagnostic number 833 (11003) . 44
Assign status of diagnostic number 834 (11004) . 45
Assign status of diagnostic number 835 (11005) . 45
Assign status of diagnostic number 841 (11006) . 45
303

IndexAssign status of diagnostic number 844 (11007) . 46
Assign status of diagnostic number 870 (11008) . 46
Assign status of diagnostic number 871 (11009) . 46
Assign status of diagnostic number 872 (11010) . 47
Assign status of diagnostic number 873 (11011) . 47
Assign status of diagnostic number 945 (11012) . 47
Assign status of diagnostic number 947 (11013) . 48
Atmospheric pressure (7601) . . . . . . . . . . . . . . . 113
Average value (0697) . . . . . . . . . . . . . . . . . . . . . 284
Average value (0698) . . . . . . . . . . . . . . . . . . . . . 283
Backlight (0111) . . . . . . . . . . . . . . . . . . . . . . . . . 26
Backup state (0121) . . . . . . . . . . . . . . . . . . . . . . . 29
Block Error
Analog input 1 to 4 (6922–1 to 4) . . . . . . . . . 165
Discrete input 1 to 2 (6857–1 to 2) . . . . . . . . 214
Block Error (10703) . . . . . . . . . . . . . . . . . . . . . . 144
Block Error (11272) . . . . . . . . . . . . . . . . . . . . . . 252
Block Error (11364) . . . . . . . . . . . . . . . . . . . . . . 243
Block tag
Analog input 1 to 4 (6901–1 to 4) . . . . . . . . . 162
Discrete input 1 to 2 (6851–1 to 2) . . . . . . . . 211
Block tag (10702) . . . . . . . . . . . . . . . . . . . . . . . 140
Block tag (11252) . . . . . . . . . . . . . . . . . . . . . . . 248
Block tag (11351) . . . . . . . . . . . . . . . . . . . . . . . 239
Calculated saturated steam pressure (1852) . . . . . 60
Calibration factor (7604) . . . . . . . . . . . . . . . . . . 119
Calorific value (7626) . . . . . . . . . . . . . . . . . . . . . . 92
Calorific value type (7698) . . . . . . . . . . . . . . . . . . 90
Calorific value unit (0552) . . . . . . . . . . . . . . . . . . 77
Calorific value unit (0606) . . . . . . . . . . . . . . . . . . 78
ChannelAnalog input 1 to 4 (6902–1 to 4) . . . . . . . . . 190
Discrete input 1 to 2 (6852–1 to 2) . . . . . . . . 228
Channel (11253) . . . . . . . . . . . . . . . . . . . . . . . . 252
Channel (11352) . . . . . . . . . . . . . . . . . . . . . . . . 243
Clear Fault State (10704) . . . . . . . . . . . . . . . . . . 151
Clear logging data (0855) . . . . . . . . . . . . . . . . . . 279
Comparison result (0103) . . . . . . . . . . . . . . . . . . . 29
Compressibility factor (7729) . . . . . . . . . . . . . . . . 64
Condensate mass flow (1857) . . . . . . . . . . . . . . . . 61
Configuration management (0100) . . . . . . . . . . . 28
Confirm Time (10705) . . . . . . . . . . . . . . . . . . . . 151
Contrast display (0105) . . . . . . . . . . . . . . . . . . . . 26
Control Totalizer 1 to 3 (0912–1 to 3) . . . . . . . . 260
Corrected volume flow (1850) . . . . . . . . . . . . . . . 57
Corrected volume flow unit (0558) . . . . . . . . . . . . 73
Corrected volume unit (0575) . . . . . . . . . . . . . . . . 74
Current
Analog input 1 to 4 (6912–1 to 4) . . . . . . . . . 199
Discrete input 1 to 2 (6842–1 to 2) . . . . . . . . 233
Current (10692) . . . . . . . . . . . . . . . . . . . . . . . . 157
Cycle Selection (10706) . . . . . . . . . . . . . . . . . . . 149
Cycle Type (10707) . . . . . . . . . . . . . . . . . . . . . . 149
Damping output (0477) . . . . . . . . . . . . . . . . . . . 130
Date/time format (2812) . . . . . . . . . . . . . . . . . . . 81
DD Resource (10708) . . . . . . . . . . . . . . . . . . . . . 146
DD Revision (10709) . . . . . . . . . . . . . . . . . 147, 272
DecimalAnalog input 1 to 4 (6961–1 to 4) . . . . . . . . . 189
304

Proline Prowirl 200 FOUNDATION Fieldbus

Analog input 1 to 4 (6980–1 to 4) . . . . . . . . . 179
Decimal places 1 (0095) . . . . . . . . . . . . . . . . . . . . 19
Decimal places 2 (0117) . . . . . . . . . . . . . . . . . . . . 20
Decimal places 3 (0118) . . . . . . . . . . . . . . . . . . . . 22
Decimal places 4 (0119) . . . . . . . . . . . . . . . . . . . . 23
Define access code (0093) . . . . . . . . . . . . . . . . . . 51
Degrees of superheat limit (7737) . . . . . . . . . . . . 49
Delta heat calculation (7736) . . . . . . . . . . . . . . . 114
Density (7607) . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Density calculation (7608) . . . . . . . . . . . . . . . . . . 88
Density unit (0555) . . . . . . . . . . . . . . . . . . . . . . . 79
Deny
Analog input 1 to 4 (6925–1 to 4) . . . . . . . . . 189
Discrete input 1 to 2 (6867–1 to 2) . . . . . . . . 227
Deny (10717) . . . . . . . . . . . . . . . . . . . . . . . . . . 147
Device address (11061) . . . . . . . . . . . . . . . . . . . 139
Device Revision (10710) . . . . . . . . . . . . . . . 147, 272
Device tag (10799) . . . . . . . . . . . . . . . . . . . . . . . 271
Device type (10711) . . . . . . . . . . . . . . . . . . 146, 272
Diagnostic event category (0738) . . . . . . . . . . . . 293
Diagnostics 1 (0692) . . . . . . . . . . . . . . . . . . . . . 265
Diagnostics 2 (0693) . . . . . . . . . . . . . . . . . . . . . 266
Diagnostics 3 (0694) . . . . . . . . . . . . . . . . . . . . . 266
Diagnostics 4 (0695) . . . . . . . . . . . . . . . . . . . . . 267
Diagnostics 5 (0696) . . . . . . . . . . . . . . . . . . . . . 268
Direct access (0106) . . . . . . . . . . . . . . . . . . . . . . . 11
Disabled
Analog input 1 to 4 (6913–1 to 4) . . . . . . . . . 200
Discrete input 1 to 2 (6843–1 to 2) . . . . . . . . 234
Disabled (10693) . . . . . . . . . . . . . . . . . . . . . . . . 157
Discrete Limit
Discrete input 1 to 2 (6863–1 to 2) . . . . . . . . 238
Discrete Priority
Discrete input 1 to 2 (6864–1 to 2) . . . . . . . . 238
Discrete Value
Discrete input 1 to 2 (6862–1 to 2) . . . . . . . . 238
Discrete Value (10746) . . . . . . . . . . . . . . . . . . . 160
Display damping (0094) . . . . . . . . . . . . . . . . . . . . 24
Display interval (0096) . . . . . . . . . . . . . . . . . . . . . 23
DSC sensor serial number (7728) . . . . . . . . . . . . 275
Dynamic viscosity (7732) . . . . . . . . . . . . . . . . . . . 93
Dynamic viscosity (7733) . . . . . . . . . . . . . . . . . . . 92
Dynamic viscosity unit (0577) . . . . . . . . . . . . . . . . 80
Energy flow (1872) . . . . . . . . . . . . . . . . . . . . . . . 61
Energy flow unit (0565) . . . . . . . . . . . . . . . . . . . . 76
Energy unit (0559) . . . . . . . . . . . . . . . . . . . . . . . . 77
ENP version (10791) . . . . . . . . . . . . . . . . . . . . . 275
Enter access code (0003) . . . . . . . . . . . . . . . . . . . 14
Enter access code (0092) . . . . . . . . . . . . . . . . . . . 14
Enthalpy calculation (7619) . . . . . . . . . . . . . . . . . 88
Enthalpy type (7620) . . . . . . . . . . . . . . . . . . . . . . 90
EU at 0%
Analog input 1 to 4 (6962–1 to 4) . . . . . . . . . 180
Analog input 1 to 4 (6981–1 to 4) . . . . . . . . . 176
EU at 100%
Analog input 1 to 4 (6963–1 to 4) . . . . . . . . . 180
Analog input 1 to 4 (6982–1 to 4) . . . . . . . . . 176
Extended order code (10795) . . . . . . . . . . . . . . . 273
Extended order code 1 (10796) . . . . . . . . . . . . . 274
Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Extended order code 2 (10797) . . . . . . . . . . . . . 274
External value (7622) . . . . . . . . . . . . . . . . . . . . 113
Failure frequency (0474) . . . . . . . . . . . . . . . . . . 132
Failure mode
Totalizer 1 to 3 (0901–1 to 3) . . . . . . . . . . . . 261
Failure mode (0451) . . . . . . . . . . . . . . . . . . . . . 131
Failure mode (0480) . . . . . . . . . . . . . . . . . . . . . 125
Failure mode (0486) . . . . . . . . . . . . . . . . . . . . . 138
Fault State (10712) . . . . . . . . . . . . . . . . . . . . . . 151
Fault State Status (11254) . . . . . . . . . . . . . . . . . 256
Fault State Status (11353) . . . . . . . . . . . . . . . . . 247
Fault State Time (11255) . . . . . . . . . . . . . . . . . . 253
Fault State Time (11354) . . . . . . . . . . . . . . . . . . 244
Fault State Val 1 (11355) . . . . . . . . . . . . . . . . . . 244
Fault State Val 2 (11356) . . . . . . . . . . . . . . . . . . 245
Fault State Val 3 (11357) . . . . . . . . . . . . . . . . . . 245
Fault State Val 4 (11358) . . . . . . . . . . . . . . . . . . 245
Fault State Val 5 (11359) . . . . . . . . . . . . . . . . . . 246
Fault State Val 6 (11360) . . . . . . . . . . . . . . . . . . 246
Fault State Val 7 (11361) . . . . . . . . . . . . . . . . . . 246
Fault State Val 8 (11362) . . . . . . . . . . . . . . . . . . 247
Fault State Value Discrete 1 (11256) . . . . . . . . . 253
Fault State Value Discrete 2 (11257) . . . . . . . . . 254
Fault State Value Discrete 3 (11258) . . . . . . . . . 254
Fault State Value Discrete 4 (11259) . . . . . . . . . 254
Fault State Value Discrete 5 (11260) . . . . . . . . . 255
Fault State Value Discrete 6 (11261) . . . . . . . . . 255
Fault State Value Discrete 7 (11262) . . . . . . . . . 255
Fault State Value Discrete 8 (11263) . . . . . . . . . 256
Feature Selection (10714) . . . . . . . . . . . . . . . . . 149
Features (10713) . . . . . . . . . . . . . . . . . . . . . . . . 148
Filter options (0656) . . . . . . . . . . . . . . . . . . . . . 269
Filter options (0705) . . . . . . . . . . . . . . . . . . . . . 269
Firmware version (10792) . . . . . . . . . . . . . . . . . 273
Fixed density (7627) . . . . . . . . . . . . . . . . . . . . . 114
Fixed process pressure (7629) . . . . . . . . . . . . . . 115
Fixed temperature (7628) . . . . . . . . . . . . . . . . . 114
Float Value
Analog input 1 to 4 (6931–1 to 4) . . . . . . . . . 207
Analog input 1 to 4 (6936–1 to 4) . . . . . . . . . 205
Analog input 1 to 4 (6946–1 to 4) . . . . . . . . . 208
Analog input 1 to 4 (6952–1 to 4) . . . . . . . . . 210
Flow damping (1802) . . . . . . . . . . . . . . . . . . . . . . 83
Flow override (1839) . . . . . . . . . . . . . . . . . . . . . . 82
Flow velocity (1865) . . . . . . . . . . . . . . . . . . . . . . 59
Format display (0098) . . . . . . . . . . . . . . . . . . . . . 16
Free Space (10715) . . . . . . . . . . . . . . . . . . . . . . 150
Free Time (10716) . . . . . . . . . . . . . . . . . . . . . . . 150
Frequency simulation (0472) . . . . . . . . . . . . . . . 290
Frequency value (0473) . . . . . . . . . . . . . . . . . . . 290
Gas mixture (7640) . . . . . . . . . . . . . . . . . . . . . . . 99
Gas type (7714) . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Grant
Analog input 1 to 4 (6926–1 to 4) . . . . . . . . . 189
Discrete input 1 to 2 (6868–1 to 2) . . . . . . . . 227
Grant (10718) . . . . . . . . . . . . . . . . . . . . . . . . . . 147
Hard Types (10719) . . . . . . . . . . . . . . . . . . . . . . 148
Hardware revision (10793) . . . . . . . . . . . . . . . . 273
Header (0097) . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Endress+Hauser

Index

Header text (0112) . . . . . . . . . . . . . . . . . . . . . . . 25
Heat flow difference (1863) . . . . . . . . . . . . . . . . . 61
High High Limit
Analog input 1 to 4 (6937–1 to 4) . . . . . . . . . 201
High High Priority
Analog input 1 to 4 (6938–1 to 4) . . . . . . . . . 201
High Limit
Analog input 1 to 4 (6939–1 to 4) . . . . . . . . . 202
High Priority
Analog input 1 to 4 (6940–1 to 4) . . . . . . . . . 202
I/O Options
Analog input 1 to 4 (6941–1 to 4) . . . . . . . . . 189
Discrete input 1 to 2 (6869–1 to 2) . . . . . . . . 227
Inlet configuration (7641) . . . . . . . . . . . . . . . . . 117
Inlet run (7642) . . . . . . . . . . . . . . . . . . . . . . . . . 117
Installation factor (7616) . . . . . . . . . . . . . . . . . . 119
Invert output signal (0470) . . . . . . . . . . . . . . . . . 138
ITK Version (10794) . . . . . . . . . . . . . . . . . . 161, 273
Language (0104) . . . . . . . . . . . . . . . . . . . . . . . . . 15
Last backup (0102) . . . . . . . . . . . . . . . . . . . . . . . 28
Length unit (0551) . . . . . . . . . . . . . . . . . . . . . . . 81
Limit Notify (10720) . . . . . . . . . . . . . . . . . . . . . 152
Linear expansion coefficient (7621) . . . . . . . . . . . 91
Linearization Type
Analog input 1 to 4 (6905–1 to 4) . . . . . . . . . 191
Locking status (0004) . . . . . . . . . . . . . . . . . . . . . 12
Logging interval (0856) . . . . . . . . . . . . . . . . . . . 278
Low Cutoff
Analog input 1 to 4 (6956–1 to 4) . . . . . . . . . 193
Low Limit
Analog input 1 to 4 (6947–1 to 4) . . . . . . . . . 203
Low Low Limit
Analog input 1 to 4 (6953–1 to 4) . . . . . . . . . 203
Low Low Priority
Analog input 1 to 4 (6954–1 to 4) . . . . . . . . . 203
Low Priority
Analog input 1 to 4 (6955–1 to 4) . . . . . . . . . 202
Manufacturer Id (10721) . . . . . . . . . . . . . . . . . . 146
Mass flow (1847) . . . . . . . . . . . . . . . . . . . . . . . . . 58
Mass flow unit (0554) . . . . . . . . . . . . . . . . . . . . . 72
Mass unit (0574) . . . . . . . . . . . . . . . . . . . . . . . . . 73
Mating pipe diameter (7648) . . . . . . . . . . . . . . . 118
Max Notify (10722) . . . . . . . . . . . . . . . . . . . . . . 152
Maximum frequency value (0454) . . . . . . . . . . . 128
Maximum value (0663) . . . . . . . . . . . . . . . . . . . 283
Maximum value (0665) . . . . . . . . . . . . . . . . . . . 284
Maximum value (7623) . . . . . . . . . . . . . . . . . . . 287
Maximum value (7633) . . . . . . . . . . . . . . . . . . . 287
Maximum value (7654) . . . . . . . . . . . . . . . . . . . 286
Maximum value (7723) . . . . . . . . . . . . . . . . . . . 285
Measuring value at maximum frequency (0475) 129
Measuring value at minimum frequency (0476) . 128
Memory Size (10723) . . . . . . . . . . . . . . . . . . . . 150
Meter body properties (7658) . . . . . . . . . . . . . . . 120
Minimum frequency value (0453) . . . . . . . . . . . . 127
Minimum value (0688) . . . . . . . . . . . . . . . . . . . 284
Minimum value (0689) . . . . . . . . . . . . . . . . . . . 283
Minimum value (7655) . . . . . . . . . . . . . . . . . . . 286
Minimum value (7724) . . . . . . . . . . . . . . . . . . . 285
305

Index

Minumum Cycle Time (10724) . . . . . . . . . . . . . . 150
Mol% Ar (7663) . . . . . . . . . . . . . . . . . . . . . . . . . 100
Mol% C2H3Cl (7664) . . . . . . . . . . . . . . . . . . . . . 100
Mol% C2H4 (7665) . . . . . . . . . . . . . . . . . . . . . . 101
Mol% C2H6 (7666) . . . . . . . . . . . . . . . . . . . . . . 101
Mol% C3H8 (7667) . . . . . . . . . . . . . . . . . . . . . . 102
Mol% CH4 (7668) . . . . . . . . . . . . . . . . . . . . . . . 102
Mol% Cl2 (7707) . . . . . . . . . . . . . . . . . . . . . . . . 102
Mol% CO (7669) . . . . . . . . . . . . . . . . . . . . . . . . 103
Mol% CO2 (7670) . . . . . . . . . . . . . . . . . . . . . . . 103
Mol% H2 (7671) . . . . . . . . . . . . . . . . . . . . . . . . 104
Mol% H2O (7672) . . . . . . . . . . . . . . . . . . . . . . . 104
Mol% H2S (7673) . . . . . . . . . . . . . . . . . . . . . . . 104
Mol% HCl (7674) . . . . . . . . . . . . . . . . . . . . . . . . 105
Mol% He (7675) . . . . . . . . . . . . . . . . . . . . . . . . 105
Mol% i-C4H10 (7676) . . . . . . . . . . . . . . . . . . . . 106
Mol% i-C5H12 (7677) . . . . . . . . . . . . . . . . . . . . 106
Mol% Kr (7678) . . . . . . . . . . . . . . . . . . . . . . . . . 106
Mol% n-C4H10 (7681) . . . . . . . . . . . . . . . . . . . . 108
Mol% n-C5H12 (7682) . . . . . . . . . . . . . . . . . . . . 108
Mol% n-C6H14 (7683) . . . . . . . . . . . . . . . . . . . . 108
Mol% n-C7H16 (7684) . . . . . . . . . . . . . . . . . . . . 109
Mol% n-C8H18 (7685) . . . . . . . . . . . . . . . . . . . . 109
Mol% n-C9H20 (7686) . . . . . . . . . . . . . . . . . . . . 109
Mol% n-C10H22 (7680) . . . . . . . . . . . . . . . . . . 107
Mol% N2 (7679) . . . . . . . . . . . . . . . . . . . . . . . . 107
Mol% Ne (7687) . . . . . . . . . . . . . . . . . . . . . . . . 110
Mol% NH3 (7688) . . . . . . . . . . . . . . . . . . . . . . . 110
Mol% O2 (7689) . . . . . . . . . . . . . . . . . . . . . . . . 111
Mol% other gas (7690) . . . . . . . . . . . . . . . . . . . 112
Mol% SO2 (7691) . . . . . . . . . . . . . . . . . . . . . . . 111
Mol% Xe (7692) . . . . . . . . . . . . . . . . . . . . . . . . . 111
Nonvolatile Cycle Time (10729) . . . . . . . . . . . . . 150
Normal mode
Analog input 1 to 4 (6958–1 to 4) . . . . . . . . . 164
Discrete input 1 to 2 (6871–1 to 2) . . . . . . . . 214
Normal mode (10726) . . . . . . . . . . . . . . . . . . . . 144
Normal mode (11265) . . . . . . . . . . . . . . . . . . . . 251
Normal mode (11367) . . . . . . . . . . . . . . . . . . . . 242
Off value low flow cutoff (1804) . . . . . . . . . . . . . . 84
On value low flow cutoff (1805) . . . . . . . . . . . . . . 84
Operating mode (0469) . . . . . . . . . . . . . . . . . . . 121
Operating time (0652) . . . . . . . . . . . . . . . . . 27, 264
Operating time from restart (0653) . . . . . . . . . . 264
Output frequency (0471) . . . . . . . . . . . . . . . 69, 132
Output State
Discrete input 1 to 2 (6874–1 to 2) . . . . . . . . 226
Permitted mode
Analog input 1 to 4 (6959–1 to 4) . . . . . . . . . 164
Discrete input 1 to 2 (6872–1 to 2) . . . . . . . . 213
Permitted mode (10727) . . . . . . . . . . . . . . . . . . 143
Permitted mode (11266) . . . . . . . . . . . . . . . . . . 251
Permitted mode (11368) . . . . . . . . . . . . . . . . . . 242
Preset value 1 to 3 (0913–1 to 3) . . . . . . . . . . . . 260
Pressure (7696) . . . . . . . . . . . . . . . . . . . . . . . . . . 63
Pressure unit (0564) . . . . . . . . . . . . . . . . . . . . . . 74
Previous diagnostics (0690) . . . . . . . . . . . . . . . . 263
Process Value Filter Time
Analog input 1 to 4 (6909–1 to 4) . . . . . . . . . 196
306

Proline Prowirl 200 FOUNDATION Fieldbus

Discrete input 1 to 2 (6855–1 to 2) . . . . . . . . 228
Pulse output (0456) . . . . . . . . . . . . . . . . . . . 68, 126
Pulse simulation (0458) . . . . . . . . . . . . . . . . . . . 290
Pulse value (0459) . . . . . . . . . . . . . . . . . . . . . . . 291
Pulse width (0452) . . . . . . . . . . . . . . . . . . . . . . 124
Reference combustion temperature (7699) . . . . . . 94
Reference density (7700) . . . . . . . . . . . . . . . . . . . 94
Reference gross calorific value (7701) . . . . . . . . . . 95
Reference pressure (7702) . . . . . . . . . . . . . . . . . . 95
Reference temperature (7703) . . . . . . . . . . . . . . . 96
Reference Z-factor (7704) . . . . . . . . . . . . . . . . . . 96
Relative density (7705) . . . . . . . . . . . . . . . . . . . . 96
Relative humidity (7731) . . . . . . . . . . . . . . . . . . 112
Relative Index
Analog input 1 to 4 (6975–1 to 4) . . . . . . . . . 197
Discrete input 1 to 2 (6886–1 to 2) . . . . . . . . 233
Relative Index (10737) . . . . . . . . . . . . . . . . . . . . 154
Reset all totalizers (2806) . . . . . . . . . . . . . . . . . 257
Reset min/max values (7706) . . . . . . . . . . . . . . . 282
Resource State (10730) . . . . . . . . . . . . . . . . . . . 145
Response time (0491) . . . . . . . . . . . . . . . . . . . . 130
Restart (10800) . . . . . . . . . . . . . . . . . . . . . . 52, 148
Reynolds number (1864) . . . . . . . . . . . . . . . . . . . 62
Reynolds number limit (7646) . . . . . . . . . . . . . . . 48
Saturated steam calculation mode (7708) . . . . . . . 89
Saturation temperature (7709) . . . . . . . . . . . . . . . 63
Select gas type (7635) . . . . . . . . . . . . . . . . . . . . . 86
Select liquid type (7636) . . . . . . . . . . . . . . . . . . . . 87
Select medium (7653) . . . . . . . . . . . . . . . . . . . . . 85
Select steam type (7637) . . . . . . . . . . . . . . . . . . . 87
Separator (0101) . . . . . . . . . . . . . . . . . . . . . . . . . 25
Serial number (10798) . . . . . . . . . . . . . . . . . . . . 272
Set Fault State (10731) . . . . . . . . . . . . . . . . . . . 152
Shed Remote Cascade (10732) . . . . . . . . . . . . . . 152
Shed Remote Out (10733) . . . . . . . . . . . . . . . . . 153
Simulate En/Disable
Analog input 1 to 4 (6966–1 to 4) . . . . . . . . . 176
Discrete input 1 to 2 (6877–1 to 2) . . . . . . . . 226
Simulate Status
Analog input 1 to 4 (6967–1 to 4) . . . . . . . . . 170
Discrete input 1 to 2 (6878–1 to 2) . . . . . . . . 220
Simulate Value
Analog input 1 to 4 (6968–1 to 4) . . . . . . . . . 173
Discrete input 1 to 2 (6879–1 to 2) . . . . . . . . 222
Simulation device alarm (0654) . . . . . . . . . . . . . 292
Simulation diagnostic event (0737) . . . . . . . . . . 293
Software option overview (0015) . . . . . . . . . . . . . 54
Specific heat capacity (7716) . . . . . . . . . . . . . . . . 97
Specific heat capacity unit (0604) . . . . . . . . . . . . . 80
Static revision
Analog input 1 to 4 (6976–1 to 4) . . . . . . . . . 197
Discrete input 1 to 2 (6887–1 to 2) . . . . . . . . 233
Static Revision
Analog input 1 to 4 (6973–1 to 4) . . . . . . . . . 162
Discrete input 1 to 2 (6884–1 to 2) . . . . . . . . 211
Static Revision (10735) . . . . . . . . . . . . . . . . . . . 140
Static revision (10738) . . . . . . . . . . . . . . . . . . . . 154
Static Revision (11270) . . . . . . . . . . . . . . . . . . . 249
Static Revision (11371) . . . . . . . . . . . . . . . . . . . 240
Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

StatusAnalog input 1 to 4 (6906–1 to 4) . . . . . . . . . 168
Analog input 1 to 4 (6923–1 to 4) . . . . . . . . . 193
Analog input 1 to 4 (6964–1 to 4) . . . . . . . . . 165
Discrete input 1 to 2 (6853–1 to 2) . . . . . . . . 217
Discrete input 1 to 2 (6865–1 to 2) . . . . . . . . 229
Discrete input 1 to 2 (6875–1 to 2) . . . . . . . . 215
Status Options
Analog input 1 to 4 (6971–1 to 4) . . . . . . . . . 190
Discrete input 1 to 2 (6882–1 to 2) . . . . . . . . 228
Status Options (11268) . . . . . . . . . . . . . . . . . . . 252
Status Options (11363) . . . . . . . . . . . . . . . . . . . 243
Steam quality (1853) . . . . . . . . . . . . . . . . . . . . . . 60
Steam quality (7605) . . . . . . . . . . . . . . . . . . . . . 116
Steam quality limit (7717) . . . . . . . . . . . . . . . . . . 49
Steam quality value (7630) . . . . . . . . . . . . . . . . . 116
Strategy
Analog input 1 to 4 (6972–1 to 4) . . . . . . . . . 162
Discrete input 1 to 2 (6883–1 to 2) . . . . . . . . 212
Strategy (10734) . . . . . . . . . . . . . . . . . . . . . . . . 141
Strategy (11269) . . . . . . . . . . . . . . . . . . . . . . . . 249
Strategy (11370) . . . . . . . . . . . . . . . . . . . . . . . . 240
SubcodeAnalog input 1 to 4 (6918–1 to 4) . . . . . . . . . 199
Analog input 1 to 4 (6928–1 to 4) . . . . . . . . . 206
Analog input 1 to 4 (6933–1 to 4) . . . . . . . . . 204
Analog input 1 to 4 (6943–1 to 4) . . . . . . . . . 208
Analog input 1 to 4 (6949–1 to 4) . . . . . . . . . 210
Discrete input 1 to 2 (6848–1 to 2) . . . . . . . . 235
Discrete input 1 to 2 (6859–1 to 2) . . . . . . . . 236
Subcode (10698) . . . . . . . . . . . . . . . . . . . . . . . . 156
Subcode (10743) . . . . . . . . . . . . . . . . . . . . . . . . 160
Switch output function (0481) . . . . . . . . . . . . . . 132
Switch output simulation (0462) . . . . . . . . . . . . 291
Switch status (0461) . . . . . . . . . . . . . . . . . . 69, 138
Switch status (0463) . . . . . . . . . . . . . . . . . . . . . 292
Switch-off delay (0465) . . . . . . . . . . . . . . . . . . . 137
Switch-off value (0464) . . . . . . . . . . . . . . . . . . . 136
Switch-on delay (0467) . . . . . . . . . . . . . . . . . . . 137
Switch-on value (0466) . . . . . . . . . . . . . . . . . . . 135
Tag Description
Analog input 1 to 4 (6974–1 to 4) . . . . . . . . . 162
Discrete input 1 to 2 (6885–1 to 2) . . . . . . . . 212
Tag Description (10736) . . . . . . . . . . . . . . . . . . 141
Tag Description (11271) . . . . . . . . . . . . . . . . . . 249
Tag Description (11372) . . . . . . . . . . . . . . . . . . 240
Target mode
Analog input 1 to 4 (6960–1 to 4) . . . . . . . . . 163
Discrete input 1 to 2 (6873–1 to 2) . . . . . . . . 212
Target mode (10728) . . . . . . . . . . . . . . . . . . . . . 141
Target mode (11267) . . . . . . . . . . . . . . . . . . . . . 250
Target mode (11369) . . . . . . . . . . . . . . . . . . . . . 241
Temperature (1851) . . . . . . . . . . . . . . . . . . . . . . 59
Temperature unit (0557) . . . . . . . . . . . . . . . . . . . 75
Terminal voltage 1 (0662) . . . . . . . . . . . . . . . . . . 68
Time Stamp
Analog input 1 to 4 (6919–1 to 4) . . . . . . . . . 198
Analog input 1 to 4 (6929–1 to 4) . . . . . . . . . 206
Analog input 1 to 4 (6934–1 to 4) . . . . . . . . . 204
Endress+Hauser

Index

Analog input 1 to 4 (6944–1 to 4) . . . . . . . . . 208
Analog input 1 to 4 (6950–1 to 4) . . . . . . . . . 209
Analog input 1 to 4 (6977–1 to 4) . . . . . . . . . 197
Discrete input 1 to 2 (6849–1 to 2) . . . . . . . . 235
Discrete input 1 to 2 (6860–1 to 2) . . . . . . . . 237
Discrete input 1 to 2 (6888–1 to 2) . . . . . . . . 232
Time Stamp (10699) . . . . . . . . . . . . . . . . . . . . . 156
Time Stamp (10739) . . . . . . . . . . . . . . . . . . . . . 154
Time Stamp (10744) . . . . . . . . . . . . . . . . . . . . . 159
Timestamp (0667) . . . . . . . . . . . . . . . . . . . . . . . 263
Timestamp (0672) . . . . . . . . . . . . . . . . . . . . . . . 264
Timestamp (0683) . . . . . . . . . . . . . . . . . . . . . . . 265
Timestamp (0684) . . . . . . . . . . . . . . . . . . . . . . . 266
Timestamp (0685) . . . . . . . . . . . . . . . . . . . . . . . 267
Timestamp (0686) . . . . . . . . . . . . . . . . . . . . . . . 267
Timestamp (0687) . . . . . . . . . . . . . . . . . . . . . . . 268
Total mass flow (1854) . . . . . . . . . . . . . . . . . . . . 60
Totalizer overflow 1 to 3 (0910–1 to 3) . . . . . . . . 67
Totalizer value 1 to 3 (0911–1 to 3) . . . . . . . . . . . 66
Transducer State
Discrete input 1 to 2 (6891–1 to 2) . . . . . . . . 226
Transducer Status
Analog input 1 to 4 (6969–1 to 4) . . . . . . . . . 173
Discrete input 1 to 2 (6880–1 to 2) . . . . . . . . 223
Transducer Value
Analog input 1 to 4 (6970–1 to 4) . . . . . . . . . 175
Discrete input 1 to 2 (6881–1 to 2) . . . . . . . . 225
Unacknowledged
Analog input 1 to 4 (6914–1 to 4) . . . . . . . . . 200
Analog input 1 to 4 (6920–1 to 4) . . . . . . . . . 198
Analog input 1 to 4 (6930–1 to 4) . . . . . . . . . 205
Analog input 1 to 4 (6935–1 to 4) . . . . . . . . . 203
Analog input 1 to 4 (6945–1 to 4) . . . . . . . . . 207
Analog input 1 to 4 (6951–1 to 4) . . . . . . . . . 209
Analog input 1 to 4 (6978–1 to 4) . . . . . . . . . 196
Discrete input 1 to 2 (6844–1 to 2) . . . . . . . . 234
Discrete input 1 to 2 (6850–1 to 2) . . . . . . . . 236
Discrete input 1 to 2 (6861–1 to 2) . . . . . . . . 237
Discrete input 1 to 2 (6889–1 to 2) . . . . . . . . 232
Unacknowledged (10694) . . . . . . . . . . . . . . . . . 157
Unacknowledged (10700) . . . . . . . . . . . . . . . . . 155
Unacknowledged (10740) . . . . . . . . . . . . . . . . . 153
Unacknowledged (10745) . . . . . . . . . . . . . . . . . 159
Unit totalizer
Totalizer 1 to 3 (0915–1 to 3) . . . . . . . . . . . . 258
Units index
Analog input 1 to 4 (6908–1 to 4) . . . . . . . . . 180
Analog input 1 to 4 (6983–1 to 4) . . . . . . . . . 176
Unreported
Analog input 1 to 4 (6915–1 to 4) . . . . . . . . . 200
Discrete input 1 to 2 (6845–1 to 2) . . . . . . . . 234
Unreported (10695) . . . . . . . . . . . . . . . . . . . . . . 157
Update State
Analog input 1 to 4 (6979–1 to 4) . . . . . . . . . 197
Discrete input 1 to 2 (6890–1 to 2) . . . . . . . . 232
Update State (10741) . . . . . . . . . . . . . . . . . . . . . 153
Value
Analog input 1 to 4 (6907–1 to 4) . . . . . . . . . 170
Analog input 1 to 4 (6921–1 to 4) . . . . . . . . . 199
307

IndexAnalog input 1 to 4 (6924–1 to 4) . . . . . . . . . 196
Analog input 1 to 4 (6965–1 to 4) . . . . . . . . . 167
Discrete input 1 to 2 (6854–1 to 2) . . . . . . . . 220
Discrete input 1 to 2 (6856–1 to 2) . . . . . . . . 236
Discrete input 1 to 2 (6866–1 to 2) . . . . . . . . 231
Discrete input 1 to 2 (6876–1 to 2) . . . . . . . . 217
Value (10701) . . . . . . . . . . . . . . . . . . . . . . . . . . 156
Value 1 display (0107) . . . . . . . . . . . . . . . . . . . . . 18
Value 2 display (0108) . . . . . . . . . . . . . . . . . . . . . 20
Value 3 display (0110) . . . . . . . . . . . . . . . . . . . . . 21
Value 4 display (0109) . . . . . . . . . . . . . . . . . . . . . 22
Value per pulse (0455) . . . . . . . . . . . . . . . . . . . . 124
Value process variable (1811) . . . . . . . . . . . . . . . 289
Velocity unit (0566) . . . . . . . . . . . . . . . . . . . . . . . 79
Volume flow (1838) . . . . . . . . . . . . . . . . . . . . . . . 57
Volume flow unit (0553) . . . . . . . . . . . . . . . . . . . 70
Volume unit (0563) . . . . . . . . . . . . . . . . . . . . . . . 72
Vortex frequency (7722) . . . . . . . . . . . . . . . . . . . 64
Write Lock (10747) . . . . . . . . . . . . . . . . . . . . . . 154
Write Priority (10748) . . . . . . . . . . . . . . . . . . . . 158
Z-factor (7631) . . . . . . . . . . . . . . . . . . . . . . . . . . 94
Direct access (Parameter) . . . . . . . . . . . . . . . . . . . . . 11
Disabled (Parameter) . . . . . . . . . . . . . . . . 157, 200, 234
Discrete input 1 to 2 (Submenu) . . . . . . . . . . . . . . . . 211
Discrete inputs (Submenu) . . . . . . . . . . . . . . . . . . . . 210
Discrete Limit (Parameter) . . . . . . . . . . . . . . . . . . . . 238
Discrete outputs (Submenu) . . . . . . . . . . . . . . . . . . . 247
Discrete Priority (Parameter) . . . . . . . . . . . . . . . . . . 238
Discrete Value (Parameter) . . . . . . . . . . . . . . . 160, 238
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Display channel 1 (Submenu) . . . . . . . . . . . . . . . . . 279
Display channel 2 (Submenu) . . . . . . . . . . . . . . . . . 280
Display channel 3 (Submenu) . . . . . . . . . . . . . . . . . 281
Display channel 4 (Submenu) . . . . . . . . . . . . . . . . . 281
Display damping (Parameter) . . . . . . . . . . . . . . . . . . 24
Display interval (Parameter) . . . . . . . . . . . . . . . . . . . 23
Document
Explanation of the structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using the document . . . . . . . . . . . . . . . . . . . . . . . . 4
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
DSC sensor serial number (Parameter) . . . . . . . . . . . 275
Dynamic viscosity (Parameter) . . . . . . . . . . . . . . . 92, 93
Dynamic viscosity unit (Parameter) . . . . . . . . . . . . . . 80

E

Energy flow (Parameter) . . . . . . . . . . . . . . . . . . . . . . 61
Energy flow unit (Parameter) . . . . . . . . . . . . . . . . . . . 76
Energy unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 77
ENP version (Parameter) . . . . . . . . . . . . . . . . . . . . . 275
Enter access code (Parameter) . . . . . . . . . . . . . . . . . . 14
Enthalpy calculation (Parameter) . . . . . . . . . . . . . . . . 88
Enthalpy type (Parameter) . . . . . . . . . . . . . . . . . . . . . 90
EU at 0% (Parameter) . . . . . . . . . . . . . . . . . . . 176, 180
EU at 100% (Parameter) . . . . . . . . . . . . . . . . . 176, 180
308

Proline Prowirl 200 FOUNDATION Fieldbus

Event list (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 270
Event logbook (Submenu) . . . . . . . . . . . . . . . . . . . . 268
Extended order code (Parameter) . . . . . . . . . . . . . . . 273
Extended order code 1 (Parameter) . . . . . . . . . . . . . 274
Extended order code 2 (Parameter) . . . . . . . . . . . . . 274
External compensation (Submenu) . . . . . . . . . . . . . 112
External pressure (Submenu) . . . . . . . . . . . . . . . . . . 287
External value (Parameter) . . . . . . . . . . . . . . . . . . . 113

F

Factory settings . . . . . . . . . . . . . . . . . . . . . . . . . . . . 294
SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 294
US units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 296
Failure frequency (Parameter) . . . . . . . . . . . . . . . . . 132
Failure mode (Parameter) . . . . . . . . 125, 131, 138, 261
Fault State (Parameter) . . . . . . . . . . . . . . . . . . . . . . 151
Fault State Status (Parameter) . . . . . . . . . . . . . 247, 256
Fault State Time (Parameter) . . . . . . . . . . . . . . 244, 253
Fault State Val 1 (Parameter) . . . . . . . . . . . . . . . . . 244
Fault State Val 2 (Parameter) . . . . . . . . . . . . . . . . . 245
Fault State Val 3 (Parameter) . . . . . . . . . . . . . . . . . 245
Fault State Val 4 (Parameter) . . . . . . . . . . . . . . . . . 245
Fault State Val 5 (Parameter) . . . . . . . . . . . . . . . . . 246
Fault State Val 6 (Parameter) . . . . . . . . . . . . . . . . . 246
Fault State Val 7 (Parameter) . . . . . . . . . . . . . . . . . 246
Fault State Val 8 (Parameter) . . . . . . . . . . . . . . . . . 247
Fault State Value Discrete 1 (Parameter) . . . . . . . . . 253
Fault State Value Discrete 2 (Parameter) . . . . . . . . . 254
Fault State Value Discrete 3 (Parameter) . . . . . . . . . 254
Fault State Value Discrete 4 (Parameter) . . . . . . . . . 254
Fault State Value Discrete 5 (Parameter) . . . . . . . . . 255
Fault State Value Discrete 6 (Parameter) . . . . . . . . . 255
Fault State Value Discrete 7 (Parameter) . . . . . . . . . 255
Fault State Value Discrete 8 (Parameter) . . . . . . . . . 256
Feature Selection (Parameter) . . . . . . . . . . . . . . . . . 149
Features (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 148
Filter options (Parameter) . . . . . . . . . . . . . . . . . . . . 269
Firmware version (Parameter) . . . . . . . . . . . . . . . . . 273
Fixed density (Parameter) . . . . . . . . . . . . . . . . . . . . 114
Fixed process pressure (Parameter) . . . . . . . . . . . . . 115
Fixed temperature (Parameter) . . . . . . . . . . . . . . . . 114
Float Value (Parameter) . . . . . . . . . . 205, 207, 208, 210
Flow damping (Parameter) . . . . . . . . . . . . . . . . . . . . 83
Flow override (Parameter) . . . . . . . . . . . . . . . . . . . . . 82
Flow velocity (Parameter) . . . . . . . . . . . . . . . . . . . . . 59
Flow velocity (Submenu) . . . . . . . . . . . . . . . . . . . . . 286
Format display (Parameter) . . . . . . . . . . . . . . . . . . . . 16
Free Space (Parameter) . . . . . . . . . . . . . . . . . . . . . . 150
Free Time (Parameter) . . . . . . . . . . . . . . . . . . . . . . 150
Frequency simulation (Parameter) . . . . . . . . . . . . . . 290
Frequency value (Parameter) . . . . . . . . . . . . . . . . . . 290
Function
see ParameterG

Gas composition (Submenu) . . . . . . . . . . . . . . . . . . . 97
Gas mixture (Parameter) . . . . . . . . . . . . . . . . . . . . . . 99
Gas type (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 99
Grant (Parameter) . . . . . . . . . . . . . . . . . . 147, 189, 227
Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

H

Hard Types (Parameter) . . . . . . . . . . . . . . . . . . . . . 148
Hardware revision (Parameter) . . . . . . . . . . . . . . . . 273
Header (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Header text (Parameter) . . . . . . . . . . . . . . . . . . . . . . 25
Heartbeat (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 287
Heat flow difference (Parameter) . . . . . . . . . . . . . . . . 61
High High Limit (Parameter) . . . . . . . . . . . . . . . . . . 201
High High Priority (Parameter) . . . . . . . . . . . . . . . . 201
High Limit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 202
High Priority (Parameter) . . . . . . . . . . . . . . . . . . . . 202

I

I/O Options (Parameter) . . . . . . . . . . . . . . . . . 189, 227
Inlet configuration (Parameter) . . . . . . . . . . . . . . . . 117
Inlet run (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 117
Installation factor (Parameter) . . . . . . . . . . . . . . . . . 119
Invert output signal (Parameter) . . . . . . . . . . . . . . . 138
IO module temperature (Submenu) . . . . . . . . . . . . . 283
ITK Version (Parameter) . . . . . . . . . . . . . . . . . 161, 273

L

Language (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 15
Last backup (Parameter) . . . . . . . . . . . . . . . . . . . . . . 28
Length unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 81
Limit Notify (Parameter) . . . . . . . . . . . . . . . . . . . . . 152
Linear expansion coefficient (Parameter) . . . . . . . . . . 91
Linearization Type (Parameter) . . . . . . . . . . . . . . . . 191
Locking status (Parameter) . . . . . . . . . . . . . . . . . . . . 12
Logging interval (Parameter) . . . . . . . . . . . . . . . . . . 278
Low Cutoff (Parameter) . . . . . . . . . . . . . . . . . . . . . . 193
Low flow cut off (Submenu) . . . . . . . . . . . . . . . . . . . . 83
Low Limit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 203
Low Low Limit (Parameter) . . . . . . . . . . . . . . . . . . . 203
Low Low Priority (Parameter) . . . . . . . . . . . . . . . . . 203
Low Priority (Parameter) . . . . . . . . . . . . . . . . . . . . . 202

M

Manufacturer Id (Parameter) . . . . . . . . . . . . . . . . . . 146
Mass flow (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 58
Mass flow unit (Parameter) . . . . . . . . . . . . . . . . . . . . 72
Mass unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 73
Mating pipe diameter (Parameter) . . . . . . . . . . . . . . 118
Max Notify (Parameter) . . . . . . . . . . . . . . . . . . . . . 152
Maximum frequency value (Parameter) . . . . . . . . . . 128
Maximum value (Parameter) . . 283, 284, 285, 286, 287
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 56
Measurement mode (Submenu) . . . . . . . . . . . . . . . . . 85
Measuring value at maximum frequency (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Measuring value at minimum frequency (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 128
Medium properties (Submenu) . . . . . . . . . . . . . . . . . 89
Medium temperature (Submenu) . . . . . . . . . . . . . . . 286
Memory Size (Parameter) . . . . . . . . . . . . . . . . . . . . 150
Meter body properties (Parameter) . . . . . . . . . . . . . 120
Min/max values (Submenu) . . . . . . . . . . . . . . . . . . . 281
Minimum frequency value (Parameter) . . . . . . . . . . 127
Minimum value (Parameter) . . . . . . 283, 284, 285, 286
Endress+Hauser

Index

Minumum Cycle Time (Parameter) . . . . . . . . . . . . . 150
Mol% Ar (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 100
Mol% C2H3Cl (Parameter) . . . . . . . . . . . . . . . . . . . 100
Mol% C2H4 (Parameter) . . . . . . . . . . . . . . . . . . . . . 101
Mol% C2H6 (Parameter) . . . . . . . . . . . . . . . . . . . . . 101
Mol% C3H8 (Parameter) . . . . . . . . . . . . . . . . . . . . . 102
Mol% CH4 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 102
Mol% Cl2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 102
Mol% CO (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 103
Mol% CO2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 103
Mol% H2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 104
Mol% H2O (Parameter) . . . . . . . . . . . . . . . . . . . . . . 104
Mol% H2S (Parameter) . . . . . . . . . . . . . . . . . . . . . . 104
Mol% HCl (Parameter) . . . . . . . . . . . . . . . . . . . . . . 105
Mol% He (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 105
Mol% i-C4H10 (Parameter) . . . . . . . . . . . . . . . . . . . 106
Mol% i-C5H12 (Parameter) . . . . . . . . . . . . . . . . . . . 106
Mol% Kr (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 106
Mol% n-C4H10 (Parameter) . . . . . . . . . . . . . . . . . . 108
Mol% n-C5H12 (Parameter) . . . . . . . . . . . . . . . . . . 108
Mol% n-C6H14 (Parameter) . . . . . . . . . . . . . . . . . . 108
Mol% n-C7H16 (Parameter) . . . . . . . . . . . . . . . . . . 109
Mol% n-C8H18 (Parameter) . . . . . . . . . . . . . . . . . . 109
Mol% n-C9H20 (Parameter) . . . . . . . . . . . . . . . . . . 109
Mol% n-C10H22 (Parameter) . . . . . . . . . . . . . . . . . 107
Mol% N2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 107
Mol% Ne (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 110
Mol% NH3 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 110
Mol% O2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 111
Mol% other gas (Parameter) . . . . . . . . . . . . . . . . . . 112
Mol% SO2 (Parameter) . . . . . . . . . . . . . . . . . . . . . . 111
Mol% Xe (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 111
Multiple analog output (Submenu) . . . . . . . . . . . . . . 239
Multiple discrete output (Submenu) . . . . . . . . . . . . . 248

N

Nonvolatile Cycle Time (Parameter) . . . . . . . . . . . . . 150
Normal mode (Parameter) . . . . 144, 164, 214, 242, 251

O

Off value low flow cutoff (Parameter) . . . . . . . . . . . . . 84
On value low flow cutoff (Parameter) . . . . . . . . . . . . . 84
Operating mode (Parameter) . . . . . . . . . . . . . . . . . . 121
Operating time (Parameter) . . . . . . . . . . . . . . . . 27, 264
Operating time from restart (Parameter) . . . . . . . . . 264
Output (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 120
Output frequency (Parameter) . . . . . . . . . . . . . . 69, 132
Output State (Parameter) . . . . . . . . . . . . . . . . . . . . 226
Output values (Submenu) . . . . . . . . . . . . . . . . . . . . . 67

P

Parameter
Structure of a parameter description . . . . . . . . . . . . 6
Permitted mode (Parameter) . . 143, 164, 213, 242, 251
Pre-amplifier temperature (Submenu) . . . . . . . . . . . 285
Preset value 1 to 3 (Parameter) . . . . . . . . . . . . . . . . 260
Pressure (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 63
Pressure unit (Parameter) . . . . . . . . . . . . . . . . . . . . . 74
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . 263
309

Index

Process parameters (Submenu) . . . . . . . . . . . . . . . . . 82
Process Value Filter Time (Parameter) . . . . . . . 196, 228
Process variables (Submenu) . . . . . . . . . . . . . . . . . . . 56
Pulse output (Parameter) . . . . . . . . . . . . . . . . . . 68, 126
Pulse simulation (Parameter) . . . . . . . . . . . . . . . . . . 290
Pulse value (Parameter) . . . . . . . . . . . . . . . . . . . . . . 291
Pulse width (Parameter) . . . . . . . . . . . . . . . . . . . . . 124
Pulse/frequency/switch output (Submenu) . . . . . . . . 120

R

Reference combustion temperature (Parameter) . . . . . 94
Reference density (Parameter) . . . . . . . . . . . . . . . . . . 94
Reference gross calorific value (Parameter) . . . . . . . . 95
Reference pressure (Parameter) . . . . . . . . . . . . . . . . . 95
Reference temperature (Parameter) . . . . . . . . . . . . . . 96
Reference Z-factor (Parameter) . . . . . . . . . . . . . . . . . 96
Relative density (Parameter) . . . . . . . . . . . . . . . . . . . 96
Relative humidity (Parameter) . . . . . . . . . . . . . . . . . 112
Relative Index (Parameter) . . . . . . . . . . . . 154, 197, 233
Reset all totalizers (Parameter) . . . . . . . . . . . . . . . . 257
Reset min/max values (Parameter) . . . . . . . . . . . . . 282
Resource block (Submenu) . . . . . . . . . . . . . . . . . . . . 140
Resource State (Parameter) . . . . . . . . . . . . . . . . . . . 145
Response time (Parameter) . . . . . . . . . . . . . . . . . . . 130
Restart (Parameter) . . . . . . . . . . . . . . . . . . . . . . 52, 148
Reynolds number (Parameter) . . . . . . . . . . . . . . . . . . 62
Reynolds number limit (Parameter) . . . . . . . . . . . . . . 48

S

Saturated steam calculation mode (Parameter) . . . . . . 89
Saturation temperature (Parameter) . . . . . . . . . . . . . 63
Select gas type (Parameter) . . . . . . . . . . . . . . . . . . . . 86
Select liquid type (Parameter) . . . . . . . . . . . . . . . . . . 87
Select medium (Parameter) . . . . . . . . . . . . . . . . . . . . 85
Select steam type (Parameter) . . . . . . . . . . . . . . . . . . 87
Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . . . 117
Sensor information (Submenu) . . . . . . . . . . . . . . . . 275
Separator (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 25
Serial number (Parameter) . . . . . . . . . . . . . . . . . . . 272
Set Fault State (Parameter) . . . . . . . . . . . . . . . . . . . 152
Shed Remote Cascade (Parameter) . . . . . . . . . . . . . . 152
Shed Remote Out (Parameter) . . . . . . . . . . . . . . . . . 153
Simulate En/Disable (Parameter) . . . . . . . . . . . 176, 226
Simulate Status (Parameter) . . . . . . . . . . . . . . 170, 220
Simulate Value (Parameter) . . . . . . . . . . . . . . . 173, 222
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 288
Simulation device alarm (Parameter) . . . . . . . . . . . . 292
Simulation diagnostic event (Parameter) . . . . . . . . . 293
Software option overview (Parameter) . . . . . . . . . . . . 54
Specific heat capacity (Parameter) . . . . . . . . . . . . . . . 97
Specific heat capacity unit (Parameter) . . . . . . . . . . . . 80
Static revision (Parameter) . . . . . . . . . . . . 154, 197, 233
Static Revision (Parameter) . . . 140, 162, 211, 240, 249
Status (Parameter) . . . . . . 165, 168, 193, 215, 217, 229
Status Options (Parameter) . . . . . . . 190, 228, 243, 252
Steam quality (Parameter) . . . . . . . . . . . . . . . . . 60, 116
Steam quality limit (Parameter) . . . . . . . . . . . . . . . . . 49
Steam quality value (Parameter) . . . . . . . . . . . . . . . 116
310

Proline Prowirl 200 FOUNDATION Fieldbus

Strategy (Parameter) . . . . . . . . 141, 162, 212, 240, 249
Subcode (Parameter) 156, 160, 199, 204, 206, 208,
210,
235,
236
Submenu
Administration . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Analog input 1 to 4 . . . . . . . . . . . . . . . . . . . . . . 161
Analog inputs . . . . . . . . . . . . . . . . . . . . . . . . . . 161
Analog outputs . . . . . . . . . . . . . . . . . . . . . . . . . 239
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . 256
Calibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 119
Communication . . . . . . . . . . . . . . . . . . . . . . . . . 139
Configuration backup display . . . . . . . . . . . . . . . . 27
Data logging . . . . . . . . . . . . . . . . . . . . . . . . . . . 275
Device information . . . . . . . . . . . . . . . . . . . . . . . 271
Diagnostic behavior . . . . . . . . . . . . . . . . . . . . . . . 32
Diagnostic handling . . . . . . . . . . . . . . . . . . . . . . . 30
Diagnostic limits . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . 265
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . 262
Discrete input 1 to 2 . . . . . . . . . . . . . . . . . . . . . . 211
Discrete inputs . . . . . . . . . . . . . . . . . . . . . . . . . . 210
Discrete outputs . . . . . . . . . . . . . . . . . . . . . . . . . 247
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Display channel 1 . . . . . . . . . . . . . . . . . . . . . . . . 279
Display channel 2 . . . . . . . . . . . . . . . . . . . . . . . . 280
Display channel 3 . . . . . . . . . . . . . . . . . . . . . . . . 281
Display channel 4 . . . . . . . . . . . . . . . . . . . . . . . . 281
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 270
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . 268
External compensation . . . . . . . . . . . . . . . . . . . . 112
External pressure . . . . . . . . . . . . . . . . . . . . . . . . 287
Flow velocity . . . . . . . . . . . . . . . . . . . . . . . . . . . 286
Gas composition . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Heartbeat . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 287
IO module temperature . . . . . . . . . . . . . . . . . . . . 283
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . 83
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 56
Measurement mode . . . . . . . . . . . . . . . . . . . . . . . 85
Medium properties . . . . . . . . . . . . . . . . . . . . . . . . 89
Medium temperature . . . . . . . . . . . . . . . . . . . . . 286
Min/max values . . . . . . . . . . . . . . . . . . . . . . . . . 281
Multiple analog output . . . . . . . . . . . . . . . . . . . . 239
Multiple discrete output . . . . . . . . . . . . . . . . . . . 248
Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 120
Output values . . . . . . . . . . . . . . . . . . . . . . . . . . . . 67
Pre-amplifier temperature . . . . . . . . . . . . . . . . . 285
Process parameters . . . . . . . . . . . . . . . . . . . . . . . 82
Process variables . . . . . . . . . . . . . . . . . . . . . . . . . 56
Pulse/frequency/switch output . . . . . . . . . . . . . . 120
Resource block . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . 117
Sensor information . . . . . . . . . . . . . . . . . . . . . . . 275
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 288
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 69
Terminal voltage . . . . . . . . . . . . . . . . . . . . . . . . 282
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 66
Totalizer 1 to 3 . . . . . . . . . . . . . . . . . . . . . . . . . . 257
Endress+Hauser

Proline Prowirl 200 FOUNDATION Fieldbus

Switch output function (Parameter) . . . . . . . . . . . . . 132
Switch output simulation (Parameter) . . . . . . . . . . . 291
Switch status (Parameter) . . . . . . . . . . . . . 69, 138, 292
Switch-off delay (Parameter) . . . . . . . . . . . . . . . . . . 137
Switch-off value (Parameter) . . . . . . . . . . . . . . . . . . 136
Switch-on delay (Parameter) . . . . . . . . . . . . . . . . . . 137
Switch-on value (Parameter) . . . . . . . . . . . . . . . . . . 135
System (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 14
System units (Submenu) . . . . . . . . . . . . . . . . . . . . . . 69

Index

Z

Z-factor (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 94

T

Tag Description (Parameter) . . . 141, 162, 212, 240, 249
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Target mode (Parameter) . . . . . 141, 163, 212, 241, 250
Temperature (Parameter) . . . . . . . . . . . . . . . . . . . . . 59
Temperature unit (Parameter) . . . . . . . . . . . . . . . . . . 75
Terminal voltage (Submenu) . . . . . . . . . . . . . . . . . . 282
Terminal voltage 1 (Parameter) . . . . . . . . . . . . . . . . . 68
Time Stamp (Parameter) 154, 156, 159, 197, 198,
204,
206,
208,
209,
232,
235,
237
Timestamp (Parameter) . . 263, 264, 265, 266, 267, 268
Total mass flow (Parameter) . . . . . . . . . . . . . . . . . . . 60
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 66
Totalizer 1 to 3 (Submenu) . . . . . . . . . . . . . . . . . . . 257
Totalizer overflow 1 to 3 (Parameter) . . . . . . . . . . . . . 67
Totalizer value 1 to 3 (Parameter) . . . . . . . . . . . . . . . 66
Transducer State (Parameter) . . . . . . . . . . . . . . . . . 226
Transducer Status (Parameter) . . . . . . . . . . . . . 173, 223
Transducer Value (Parameter) . . . . . . . . . . . . . 175, 225

U

Unacknowledged (Parameter) 153, 155, 157, 159,
196, 198, 200, 203, 205, 207, 209, 232, 234, 236, 237
Unit totalizer (Parameter) . . . . . . . . . . . . . . . . . . . . 258
Units index (Parameter) . . . . . . . . . . . . . . . . . 176, 180
Unreported (Parameter) . . . . . . . . . . . . . 157, 200, 234
Update State (Parameter) . . . . . . . . . . . . 153, 197, 232

V

Value (Parameter) 156, 167, 170, 196, 199, 217,
220,
231,
236
Value 1 display (Parameter) . . . . . . . . . . . . . . . . . . . 18
Value 2 display (Parameter) . . . . . . . . . . . . . . . . . . . 20
Value 3 display (Parameter) . . . . . . . . . . . . . . . . . . . 21
Value 4 display (Parameter) . . . . . . . . . . . . . . . . . . . 22
Value per pulse (Parameter) . . . . . . . . . . . . . . . . . . 124
Value process variable (Parameter) . . . . . . . . . . . . . 289
Velocity unit (Parameter) . . . . . . . . . . . . . . . . . . . . . 79
Volume flow (Parameter) . . . . . . . . . . . . . . . . . . . . . 57
Volume flow unit (Parameter) . . . . . . . . . . . . . . . . . . 70
Volume unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 72
Vortex frequency (Parameter) . . . . . . . . . . . . . . . . . . 64

W

WizardDefine access code . . . . . . . . . . . . . . . . . . . . . . . . 50
Write Lock (Parameter) . . . . . . . . . . . . . . . . . . . . . . 154
Write Priority (Parameter) . . . . . . . . . . . . . . . . . . . . 158

Endress+Hauser311

www.addresses.endress.com