Installation Instructions
OXY5500 Transportable Analyzer
EA01389C/66/EN/01.21

The following instructions are being provided for the OXY5500 Transportable
analyzer configuration. Refer to the manuals on the enclosed System
Documents CD for additional programming information, analyzer installationand maintenance guidelines.
The OXY5500 Transportable analyzer contains the analyzer and modified
sample conditioning panel secured inside a rolling Pelican case. Thisconfiguration is not intended for permanent installation, but for periodic use atmultiple locations.

Connecting electrical power to the analyzer
The OXY5500 Transportable analyzer is powered by 9 to 36 VDC
or 120 to 240 VAC. Refer to the Analyzer Operator’s manuals
provided with the shipment for additional installation information.

1. Open the lid to the portable carrying case. The OXY5500 analyzer
and sample conditioning system (SCS) are mounted to the inside.
2. Open the SS2100 analyzer electronics enclosure door. Take care not
to disturb the electrical assembly inside.
Hazardous voltage and risk of electric shock. Failure toproperly ground the analyzer may create a high-voltage shockhazard.

3. Remove the red plug caps from the conduit hub on the right side of
the analyzer enclosure labeled for power input.
4. Run armor-braided cable to the conduit hub.
Cable gland should be used where appropriate in compliance withlocal regulations.

Endress+Hauser

1

OXY5500 Transportable Analyzer

EA01389C

The electrical installation to which the apparatus is connectedmust be protected against transients. The protective device has
to be set at a level not exceeding 140% of the peak rated voltage
values at the power supply terminals.

5. For AC systems, pull ground, neutral (N) and L1 wires into theelectronics enclosure. Refer to Figure 1.
For DC power, pull VI +, – and ground wires into the electronics
enclosure. Refer to Figure 1.

DC LO
DC ONNV1 +AC POWER CONNECTION–DC POWER CONNECTION

Figure 1 AC/DC power connections

6. Strip back the jacket and/or insulation of the wires just enough toconnect to the power terminal block.
7. Connect the main ground wire to the protection ground terminal
marked.
8. Close and tighten the analyzer enclosure door.
Apply 4 N-m (35 in-lbs) of torque on each bolt to ensure the door
is closed properly to maintain required ingress protection.

2Endress+Hauser

EA01389C

OXY5500 Transportable Analyzer

Sample Conditioning System (SCS) Panel
The SCS has received minor modifications from the standard configuration
provided with the OXY5500. The differences are most apparent in the parts
used rather than the operation. Refer to the OXY5500 Sample Conditioning
System (SCS) Overview Manual (P/N 4900002244) for information on the SCS
operation and maintenance.

Spare Parts
The following spare parts list is specific to the OXY5500 Transportable analyzer
SCS only. Other parts for this configuration type can be found in the OXY5500
Sample Conditioning System (SCS) Overview Manual (P/N 4900002244). Referto the OXY5500 Operator’s Manual (P/N 4900002239) provided with theshipment for information regarding spare parts for the analyzer. Whenordering, please specify the system serial number (SN) to ensure that thecorrect parts are identified.
Table 1 Replacement parts for OXY5500 Transportable SCS

Part
NumberDescription

2 YR
QTY

Sample Conditioning System
6101671208

Needle Valve, 1/4 in. TF (SS)

-6100002004

Pressure Gauge (1.5 in.), 30 PSIG, 1/8 in. MNPT (SS)

-6100322140

Quick Disconnect, Deso Stem, 1/4 x 1/4 sw, SS316-Service Contact
For Service, refer to our website (www.endress.com) for the list of local sales
channels in your area.

Endress+Hauser

3

OXY5500 Transportable Analyzer

EA01389C

THIS PAGE INTENTIONALLY LEFT BLANK

4Endress+Hauser