COMPATIBLE DISPLAYS & PC INTERFACES

MACH6KEY FEATURES)APM

ACCESSORIESUP TO 200 kHz PULSE-TO-PULSE

Measure EVERY pulse, with no sampling, at high rep rates,up to 200 kHz)CAPTURE AND STORE UP TO 4 MILLION PULSESStore 40 seconds of data at 100 kHz)TRACK MISSING PULSES AND PULSES BELOWTHRESHOLD

Stand with delrin post

Additional 9V power supply

USB cable

APM analog power supply

M6-UV-QED
Relative measurements in UV

Pelican carrying case

Know how many pulses were missed or that didn't makethe energy threshold with this unique pulse feature)

SEVERAL HEADS TO CHOOSE FROMSilicon, lnGaAs and pyroelectric heads for a broad
wavelength and energy range)

ANALOG MODULE AVAILABLE

Use our fast M6 Detectors with the APM and anoscilloscope for fast analog energy measurements)

FULL-SPEED USB 2.0 CONNECTION

Ensures high data rate transfer and fast operation)

USER-FRIENDLY SOFTWARE WITH MANYDIAGNOSTIC FEATURES• Live mode, strip chart, histogram and statistics displays• FFT display of pulse energy data for temporal
diagnostics• Life test mode to automate laser testinggentec-eo.com / laser-energy-meter

Nisr· r:-7
CE TraceableE�!i';;�
���'Also traceable to NRC-CNRC

Specifications.
".
M6-6-SI

M6-6-SI-L

M6-6-IN

M6-6-IN-L

M6-6-PV

M6-12.5-PV

MAX ENERCY8200µJ

2 µJ

200µJ

2 µJ

2mJ

20mJ

MAXAVERACE POWER

SW

SW

SW

SW

SW

25W

MAX REP RATE

200 000 Hz

200000 Hz

200000 Hz

200000 Hz

200 000 Hz

200 000 Hz

EFFECTIVE APERTURE

6mm(/)

6mm(/)

6mm(/)

6mm(/)

6mm(/)

12.Smm(/)

Spectral range

0.35-1.l µm

0.35-1.l µm

0.9-1.6 µm

0.9-1.6 µm

0.35-2.5 µm

0.35-2.5 µm

Max measurable energy'

200µJ

2 µJ

200µJ

2 µJ

2mJ

20mJ

MEASUREMENT CAPABILITY

Noise equivalent energy

2 nJ

20pJ

2 nJ

20pJ

0.2 µJ

0.2µJ

Rise time (□-100%)150ns

150 ns

150ns

150ns

150ns

150ns

Max repetition rate

200 000 Hz

200000 Hz

200000 Hz

200000 Hz

200 000 Hz

200 000 Hz

Max pulse width

100 ns

lOOns

lOOns

lOOns

lOOns

lOOns

Calibration uncertainty

±4%

±4%

±4%

±4%±4%±4%

Repeatability

±1%

±1%

±1%

±1%

±1%

±1%

Max average power (60 seconds)

SW

SW

SW

SW

SW

25W

Max energy (at 1064 nm)

200µJ

2 µJ

200µJ

2 µJ

20mJ

20mJ
12.Smm(/);o(/lDAMACE THRESHOLDSPHYSICAL CHARACTERISTICS

Effective aperture

6mm(/)

6mm(/)

6mm(/)

6mm(/)

6mm(/)

SensorSiliconSiliconlnGaAslnGaAsPyroelectric

Pyroelectric

Dimensions

38.l (/) x 58.4D mm

38.l (/) x 58.4D mm

38.l (/) x 58.4D mm

38.l (/) x 58.4D mm

38.l (/) x 58.4D mm

76Hxll1Wx76D

Weight (head only)

150g

150g

150g

150g

150g

N/A

STAND-D-233

STAND-D-233

COMPATIBLE METERS

PC-based

Mach 6: see detailed specifications on next page

Analog power supply

APM: see detailed specifications on next page

ORDERINC INFORMATION

Compatible standProduct page•

STAND-D-233

STAND-D-233•

STAND-D-233(!I_[!I

STAND-D-233.

a. Maximum pulse energy reading will vary with wavelength and probe voltage responsivity (Rv). For more details, please read Application Note 1210-201932 and contact us at info@gentec-eo.com

PRODUCT GUIDE 2022

Specifications are subject to change without notice
T 418.657.8003 I info@gentec-eo.com

genteC-C•J.com

75Specifications

STRIP CHART
The strip chart display provides a quick graphical look at the pulse data batch
just collected. The data can be displayed in full scale or in auto scale mode.
You can also zoom-in on a portion of the data, like shown in the screen on
the left. An ND0.3 filter has been dropped through the beam and you can see
the effects on the pulse data collected. You can fit trend and min/max lines
to the data. Just below the chart, you will find a complete set of statistics forthe batch. At the top of this screen you will see tabs that will take you to theHistogram, Statistics, and FFT (Fast Fourier Transform) displays.

1--,.,.,......

--... -... --1""■-,---.,._ __HISTOGRAM
Interested in viewing the statistical distribution of the pulse energy data
set? The Histogram screen does this for you and fits a "best Gaussian curve"
to the data. It displays complete statistical calculations along the bottom ofthe graph, along with pulse frequency. Just below the statistics, you will findinstrument controls, like range and trigger. You are also given the ability toadjust trigger delay and hold off as needed.

STATISTICS
The statistics display offers a very complete set of useful energy readings andcalculated statistics. These include: minimum, maximum, average (mean),standard deviation, spread and variation. Some other very handy features
include: windows displaying, average frequency, pulse jitter, pulses below trigger
and pulses below threshold (a level set by you). In the screen on the left, you
can see that there were 72 pulses below an energy threshold of 3 µW, and 747
missing pulses (or pulses below trigger).

LIFE TEST MODE=--::..1..
,_,,..,.

-.

.

---

---•fM(l' ft. .ild!ld]lll,;.,,,.uu.-J"

1. . . . . . . U6£} . . . 11!41Ui.al'i::it:-"'-..,..IIIIH{ --.
-PRODUCT GUIDE 2022

Would you like to run a life test on your high repetition rate pulsed laser? Howabout a periodic test vs. an environmental change like temperature? We haveincluded a great feature to accomplish this. In the life test screen, we give you
the ability to select the statistics you want, a start time and date, a stop timeand date, the number of pulses and a test interval. You simply identify a file, aplace to put the data, and then click on start and walk away! When you comeback, you have a data set that tracked the performance of your laser over time,
temperature, shock, vibration or anything you chose.

T 418.657.8003 I info@gentec-eo.com〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068,

genteC-EO
https://www.phototechnica.co.jp
e-mail:voc@phototechnica.co.jp

77