••••••••

DIMENSIONSZZZSKRWRWHFKQLFDFRMSࠛᇸ⋢┴ࡉ࠸ࡓࡲᕷ༡༊༡ᾆ࿴ࠉ7(/)$;
HPDLOYRF#SKRWRWHFKQLFDFRMSNEXOSTM SPECTROMETERTECHNICAL DATASPECIFICATIONSNEXOSTM 2K

NEXOSTM 4K

Symmetrical Czerny-Turner design, 75 mm focal length; NEXOS

Optical Bench

TM

Spectrometer bench

190 - 1100 nm

Wavelength range

0.1 - 1% (typical value 300l/mm, blaze 300 nm < 0.3%)

Stray light
Detector

HAM S11639, CMOS linear array, 2048 pixels (14x200 µm)

HAM S13496, CMOS linear array, 4096 pixels (7x200µm)

375:1

365:1

Signal/Noise

4500

Dynamic Range
Dark noise

15 cnts

AD converter

16-bit, 6 MHz
9 µs – 30 s

Integration timeUSB 2.0 (480Mbps) / pigtailed (38cm) USB-AInterface
Sample speed on-board averaging

0.36 ms/scan

0.70 ms/scan

Data transfer speed

0.79 ms/scan

1.12 ms/scan

5 bidirectional programmable I/O; 1 Analog out, 1 Analog in, 1x5V

Digital I/O
Dimensions, weight

105 x 80 x 20 mm, 277,5 grams

Power supply

Default USB power, 500 mA
5-55 °C

Temperature range

GRATINGS
Use

Useable range (nm)Spectral range (nm)Lines/mm

Blaze (nm)Order code

UV/VIS/NIR

190-1160**

910**

300

300

MN0300-0.30

UV/VIS

190-850544-540*600

300

MN0600-0.30

UV

190-750

261-256*

1200

250

MN1200-0.25

UV

190-650

168-152*

1800

UV

MN1800-0.25

UV

190-580

121-103*

2400

UV

MN2400-0.25

UV

190-400

72-57*3600

UV

MN3600-0.25

UV/VIS

250-850536-532*600

400

MN0600-0.40

VIS/NIR

300-1160**

800**

300

500

MN0300-0.50

VIS

360-1000

529-519*600

500

MN0600-0.50

VIS

300-800

247-216*

1200

500

MN1200-0.50

VIS

350-750147-121*

1800

500

MN1800-0.50

VIS

350-640

96-89*2400

VIS

MN2400-0.50

NIR500-1050

515-506*

600

750

MN0600-0.75

NIR

500-1050

210-204*

1200

750

MN1200-0.75

NIR

600-1160

339-300*

830

800

MN0830-0.80

NIR

600-1160**

500**

300

1000

MN0300-1.00

NIR

600-1160

500

600

1000

MN0600-1.00

Note: a selection of starting wavelength is possible, contact us for available configurations.
* Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and smaller the range to select** Please note that not all pixels will be used for the useable range

MORE DATA ON THE NEXT PAGE >RESOLUTION NEXOSTM 2K
Slit size (µm)

102550100200500

300 lines/mm grating

1.0

1.4

2.25

4.8

9.2

21.3

600 lines/mm grating

0.40-0.53*

0.7

1.2

2.4

4.6

10.8

830 lines/mm grating

0.32

0.48

0.93

1.7

3.4

8.5

1200 lines/mm grating

0.20-0.28*

0.27-0.38*

0.52-0.66*

1.1

2.3

5.4

1800 lines/mm grating

0.10-0.18*

0.20-0.29*

0.34-0.42*

0.8

1.6

3.6

2400 lines/mm grating

0.09-0.13*

0.13-0.17*

0.26-0.34*

0.44-0.64*

1.1

2.7

3600 lines/mm grating

0.06-0.08*

0.10

0.19

0.4

0.8

1.8

*Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and the higher the resolution.

RESOLUTION NEXOSTM 4K
Slit size (µm)

102550100200500

300 lines/mm grating

0.50-0.70

1.4

2.25

4.8

9.2

21.3

600 lines/mm grating

0.30-0.36*

0.7

1.2

2.4

4.6

10.8

830 lines/mm grating

0.25

0.48

0.93

1.7

3.4

8.5

1200 lines/mm grating

0.14-0.18*

0.27-0.38*

0.52-0.66*

1.1

2.3

5.4

1800 lines/mm grating

0.09-0.11*

0.20-0.29*

0.34-0.42*

0.8

1.6

3.6

2400 lines/mm grating

0.07-0.09*

0.13-0.17*

0.26-0.34*

0.44-0.64*

1.1

2.7

3600 lines/mm grating

0.05-0.06*

0.10

0.19

0.4

0.8

1.8

*Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and the higher the resolution.

ORDERING INFORMATION
Order code

Ordering information

AvaSpec-NXS2048CL

NEXOSTM fiber-optic spectrometer, 75 mm focal length, 2048 pixel CMOS detector, USB 2 powered interface, needs DCL, secondorder suppressing, replaceable slit

AvaSpec-NXS4096CL

NEXOSTM fiber-optic spectrometer, 75 mm focal length, 4096 pixel CMOS detector, USB 2 powered interface, needs DCL, second
order suppressing, replaceable slit

*OEM AvaSpecNXS2048CL

OEM NEXOSTM fiber-optic spectrometer, 75 mm focal length, 2048 pixel CMOS detector, USB 2 powered interface, needs DCL,
second order suppressing, fixed slit

*OEM AvaSpecNXS4096CLOEM NEXOSTM fiber-optic spectrometer, 75 mm focal length, 4096 pixel CMOS detector, USB 2 powered interface, needs DCL,
second order suppressing, fixed slit

* Only for OEM customers

CONFIGURATION DEPENDANT PARAMETERS
Order code

Parameter information
spectrometer for non OSF configurations. Slit size, please specify XX = 10, 25, 50, 100, 200, 500 µm

SLIT-XX-PRS

Replaceable Slit for NEXOS

SLIT-XX-PRSF

Replaceable Slit for NEXOSTM spectrometer for OSF configurations. Slit size, please specify XX = 10, 25, 50, 100, 200, 500 µm*SLIT-XX-P

Fixed Slit for NEXOSTM spectrometer. Slit size, please specify XX = 10, 25, 50, 100, 200, 500 µm

OSC

Order sorting coating for grating MN 0600-0.50

OSC-UA

Order sorting coating for grating MN 0300-0.30 / MN 0300-0.50

OSC-UB

Order sorting coating for grating MN 0600-0.50

OSF-YYY-3

Order sorting filter for reduction of second-order effects, possible: YYY = 305, 395, 457, 515, 550, or 600 nm,
depending on range

DCL

Detector Collection Lens. To be included

TM* Only for OEM customers
Avantes will select needed second order suppression based on desired configuration.

Discover all our empowering spectroscopy solutions on our website www.avantes.com
We reserve the right to make changes without prior notice. While every effort has been made to ensure all
information in this document is accurate, Avantes accepts no liability for any inaccuracies.

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068

e-mail:voc@phototechnica.co.jp

Version 1 - 14.12

NEXOSTM LINKTECHNICAL DATASPECIFICATIONSNEXOSTM 2K
Symmetrical Czerny-Turner design, 75 mm focal length; NEXOS

Optical Benchspectrometer bench

190 - 1100 nm

Wavelength range

0.1 - 1% (typical value 300l/mm, blaze 300 nm < 0.3%)

Stray lightDetector

NEXOSTM 4KTM

HAM S11639, CMOS linear array, 2048 pixels (14x200 µm)

HAM S13496, CMOS linear array, 4096 pixels (7x200µm)

375:1

365:1

Signal/Noise

4500

Dynamic Range
Dark noise

15 cnts

AD converter

16-bit, 6 MHz

Integration time

9 µs – 30 s
RS232 or SPI pigtailed with separate power connection

Interface

TBD ms /scan

Data transfer speed

TBD ms /scan

5 bidirectional programmable I/O; 1 Analog out, 1 Analog in, 1x5V

Digital I/O

105 x 80 x 20 mm, 277,5 grams

Dimensions, weight
Power supply

Separate

Temperature range

5-55 °CGRATINGS
Use

Useable range (nm)Spectral range (nm)Lines/mm

Blaze (nm)Order code

UV/VIS/NIR

190-1160**

910**

300

300

MN0300-0.30

UV/VIS

190-850544-540*600

300

MN0600-0.30

UV

190-750

261-256*

1200

250

MN1200-0.25

UV

190-650

168-152*

1800

UV

MN1800-0.25

UV

190-580

121-103*

2400

UV

MN2400-0.25

UV

190-400

72-57*3600

UV

MN3600-0.25

UV/VIS

250-850536-532*600

400

MN0600-0.40

VIS/NIR

300-1160**

800**

300

500

MN0300-0.50

VIS

360-1000

529-519*600

500

MN0600-0.50

VIS

300-800

247-216*

1200

500

MN1200-0.50

VIS

350-750

147-121*

1800

500

MN1800-0.50

VIS

350-640

96-89*2400

VIS

MN2400-0.50

NIR500-1050

515-506*

600

750

MN0600-0.75

NIR

500-1050

210-204*

1200

750

MN1200-0.75

NIR

600-1160

339-300*

830

800

MN0830-0.80

NIR

600-1160**

500**

300

1000

MN0300-1.00

NIR

600-1160

500

600

1000

MN0600-1.00

Note: a selection of starting wavelength is possible, contact us for available configurations.
* Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and smaller the range to select** Please note that not all pixels will be used for the useable range

MORE DATA ON THE NEXT PAGE >RESOLUTION NEXOSTM 2K
Slit size (µm)

102550100200500

300 lines/mm grating

1.0

1.4

2.25

4.8

9.2

21.3

600 lines/mm grating

0.40-0.53*

0.7

1.2

2.4

4.6

10.8830 lines/mm grating

0.32

0.48

0.93

1.7

3.4

8.5

1200 lines/mm grating

0.20-0.28*

0.27-0.38*

0.52-0.66*

1.1

2.3

5.4

1800 lines/mm grating

0.10-0.18*

0.20-0.29*

0.34-0.42*

0.8

1.6

3.6

2400 lines/mm grating

0.09-0.13*

0.13-0.17*

0.26-0.34*

0.44-0.64*

1.1

2.7

3600 lines/mm grating

0.06-0.08*

0.10

0.19

0.4

0.8

1.8

*Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and the higher the resolution.

RESOLUTION NEXOSTM 4K
Slit size (µm)

102550100200500

300 lines/mm grating0.50-0.70

1.4

2.25

4.8

9.2

21.3

600 lines/mm grating

0.30-0.36*

0.7

1.2

2.4

4.6

10.8

830 lines/mm grating

0.25

0.48

0.93

1.7

3.4

8.5

1200 lines/mm grating

0.14-0.18*

0.27-0.38*

0.52-0.66*

1.1

2.3

5.4

1800 lines/mm grating

0.09-0.11*

0.20-0.29*

0.34-0.42*

0.8

1.6

3.6

2400 lines/mm grating

0.07-0.09*

0.13-0.17*

0.26-0.34*

0.44-0.64*

1.1

2.7

3600 lines/mm grating

0.05-0.06*

0.10

0.19

0.4

0.8

1.8

*Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and the higher the resolution.

ORDERING INFORMATION
Order code

Ordering information

AvaSpec-NXS2048CL-RS232

OEM NEXOS fiber-optic spectrometer, 75 mm focal length, 2048 pixel CMOS detector, RS232 communication, power connection, needs DCL, second order suppressing, fixed slit

AvaSpec-NXS4096CL-RS232

OEM NEXOSTM fiber-optic spectrometer, 75 mm focal length, 4096 pixel CMOS detector, RS232 communication, power connection, needs DCL, second order suppressing, fixed slit

AvaSpec-NXS2048CL-SPI

OEM NEXOSTM fiber-optic spectrometer, 75 mm focal length, 2048 pixel CMOS detector, SPI communication, power
connection, needs DCL, second order suppressing, fixed slit

AvaSpec-NXS4096CL-SPI

OEM NEXOSTM fiber-optic spectrometer, 75 mm focal length, 4096 pixel CMOS detector, SPI communication, power
connection, needs DCL, second order suppressing, fixed slit

TM

CONFIGURATION DEPENDANT PARAMETERS
Order code

Options information

SLIT-XX-P

Fixed Slit for NEXOSTM Spectrometer. Slit size, please specify XX = 10, 25, 50, 100, 200, 500 µm

OSC

Order sorting coating for grating MN 0600-0.50

OSC-UA

Order sorting coating for grating MN 0300-0.30 / MN 0300-0.50

OSC-UB

Order sorting coating for grating MN 0600-0.50

OSF-YYY-3

Order sorting filter for reduction of second-order effects, possible: YYY = 305, 395, 457, 515, 550, or 600 nm,depending on range

DCL

Detector Collection Lens. To be included

Avantes will select needed second order suppression based on desired configuration.

Discover all our empowering spectroscopy solutions on our website www.avantes.comWe reserve the right to make changes without prior notice. While every effort has been made to ensure all
information in this document is accurate, Avantes accepts no liability for any inaccuracies.

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067 FAX:048-871-0068
e-mail:voc@phototechnica.co.jp

Version 1 - 14.12

NEXOSTM BENCH ONLYTECHNICAL DATASPECIFICATIONSNEXOSTM 2K

NEXOSTM 4K

Symmetrical Czerny-Turner design, 75 mm focal length; NEXOSTM Spectrometer bench

Optical Bench

190 - 1100 nm

Wavelength range

0.1 - 1% (typical value 300l/mm, blaze 300 nm < 0.3%)

Stray light

HAM S11639, CMOS linear array, 2048 pixels (14x200 µm)

Detector

HAM S13496, CMOS linear array, 4096 pixels (7x200µm)

105 x 80 x 20 mm, 277,5 grams

Dimensions, weight5-55 °C

Temperature range

Direct interfacing to connector via ZIF and coax cable

InterfaceGRATINGS
Useable range (nm)Spectral range (nm)Lines/mm

Blaze (nm)Order code

UV/VIS/NIR

Use

190-1160**

910**

300

300

MN0300-0.30

UV/VIS

190-850544-540*600

300

MN0600-0.30

UV

190-750

261-256*

1200

250

MN1200-0.25

UV

190-650

168-152*

1800

UV

MN1800-0.25

UV

190-580

121-103*

2400

UV

MN2400-0.25

UV

190-400

72-57*3600

UV

MN3600-0.25

UV/VIS

250-850536-532*600

400

MN0600-0.40

VIS/NIR

300-1160**

800**

300

500

MN0300-0.50

VIS

360-1000

529-519*600

500

MN0600-0.50

VIS

300-800

247-216*

1200

500

MN1200-0.50

VIS

350-750

147-121*

1800

500

MN1800-0.50

VIS

350-640

96-89*2400

VIS

MN2400-0.50

NIR500-1050

515-506*

600

750

MN0600-0.75

NIR

500-1050

210-204*

1200

750

MN1200-0.75NIR

600-1160

339-300*

830

800

MN0830-0.80

NIR

600-1160**

500**

300

1000

MN0300-1.00

NIR

600-1160

500

600

1000

MN0600-1.00

Note: a selection of starting wavelength is possible, contact us for available configurations.
* Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and smaller the range to select** Please note that not all pixels will be used for the useable range

MORE DATA ON THE NEXT PAGE >RESOLUTION NEXOSTM 2KSlit size (µm)

102550

100

200

500

300 lines/mm grating

1.01.42.25

4.8

9.2

21.3

600 lines/mm grating

0.40-0.53*

0.7

1.2

2.4

4.6

10.8

830 lines/mm grating

0.32

0.48

0.93

1.7

3.4

8.5

1200 lines/mm grating

0.20-0.28*

0.27-0.38*

0.52-0.66*

1.1

2.3

5.4

1800 lines/mm grating

0.10-0.18*

0.20-0.29*

0.34-0.42*

0.8

1.6

3.6

2400 lines/mm grating

0.09-0.13*

0.13-0.17*

0.26-0.34*

0.44-0.64*

1.1

2.7

3600 lines/mm grating

0.06-0.08*

0.10

0.19

0.4

0.8

1.8

*Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and the higher the resolution.

RESOLUTION NEXOSTM 4K
Slit size (µm)

102550100

200500300 lines/mm grating

0.50-0.70

1.4

2.25

4.8

9.2

21.3

600 lines/mm grating

0.30-0.36*

0.7

1.2

2.4

4.6

10.8

830 lines/mm grating

0.25

0.48

0.93

1.7

3.4

8.5

1200 lines/mm grating

0.14-0.18*

0.27-0.38*

0.52-0.66*

1.1

2.3

5.4

1800 lines/mm grating

0.09-0.11*

0.20-0.29*

0.34-0.42*

0.8

1.6

3.6

2400 lines/mm grating0.07-0.09*

0.13-0.17*

0.26-0.34*

0.44-0.64*

1.1

2.7

3600 lines/mm grating

0.05-0.06*

0.10

0.19

0.4

0.8

1.8

*Depends on the starting wavelength of the grating; the higher the wavelength, the bigger the dispersion and the higher the resolution.

ORDERING INFORMATION
Order code

Ordering information

AvaBench-75NXS2048CL-U3

OEM NEXOS fiber-optic spectrometer bench, 75 mm focal length, 2048 pixel CMOS detector, needs DCL, second order suppressing, fixed slit

AvaBench-75NXS4096CL-U3

OEM NEXOSTM fiber-optic spectrometer bench, 75 mm focal length, 4096 pixel CMOS detector, needs DCL, second order suppressing, fixed slit

TM

CONFIGURATION DEPENDANT PARAMETERS
Order code

Parameter information

SLIT-XX-P

Fixed Slit for NEXOSTM Spectrometer. Slit size, please specify XX = 10, 25, 50, 100, 200, 500 µm

OSC

Order sorting coating for grating MN 0600-0.50

OSC-UA

Order sorting coating for grating MN 0300-0.30 / MN 0300-0.50

OSC-UB

Order sorting coating for grating MN 0600-0.50

OSF-YYY-3

Order sorting filter for reduction of second-order effects, possible: YYY = 305, 395, 457, 515, 550, or 600 nm,depending on range

DCL

Detector Collection Lens. To be included

Avantes will select needed second order suppression based on desired configuration.

Discover all our empowering spectroscopy solutions on our website www.avantes.com
We reserve the right to make changes without prior notice. While every effort has been made to ensure all
information in this document is accurate, Avantes accepts no liability for any inaccuracies.

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068

e-mail:voc@phototechnica.co.jp

Version 1 - 14.12