Development of a head-mounted, eye-tracking system for dogs

Abstract
Growing interest in canine cognition and visual perception has promoted research
into the allocation of visual attention during free-viewing tasks in the dog. The
techniques currently available to study this (i.e. preferential looking) have, however,lacked spatial accuracy, permitting only gross judgements of the location of the dog’s
point of gaze and are limited to a laboratory setting. Here we describe a mobile, headmounted, video-based, eye-tracking system and a procedure for achieving
standardised calibration allowing an output with accuracy of 2-3º.

The setup allows free movement of dogs; in addition the procedure does not involve
extensive training skills, and is completely non-invasive. This apparatus has thepotential to allow the study of gaze patterns in a variety of research applications andcould enhance the study of areas such as canine vision, cognition and social
interactions.

Keywords: Attention; dogs; eye tracking; gaze; mobility.

1. Introduction
Visual exploration of our environment involves a series of saccades (along with headand body movements, which dogs frequently make) to direct our gaze to regions
either informative or interesting to us. The preferred regions within a scene are often
inspected earlier and attract more fixations and longer viewing time. Gaze patterns
hence provide a real-time behaviour index of ongoing perceptual and cognitive1

processing, and could be sensitive indices of our attention, motivation and preference,
especially when exploring scenes of high ecological validity (Rayner, 1998;
Henderson, 2003; Land et al., 2006).

To monitor human gaze patterns, scleral search coils, dual Purkinje image eye
trackers and video-based, pupil-centre/corneal-reflection eye trackers are used indifferent laboratory settings for different requirements of spatial and temporal
resolution. Among these different eye tracking devices, only video-based, headmounted systems allow free head and body movements, and can be used to study
naturalistic vision in everyday activities such as driving, playing sports and preparing
food (Land et al., 2006).

The adaption of eye-tracking systems for use on non-human animals has, however,resulted in the use of a number of divergent techniques to measure eye movements
varying in invasiveness, restraint and level of training required for the use of theapparatus.

With the highest spatial and temporal resolution, scleral search coils are typically
employed to study visual processing and eye-movement control in non-humanprimates, such as macaque monkeys. However the protocol is invasive requiring
surgical implantation of a scleral magnetic search coil under the conjunctiva aroundthe eyeball (Judge et al., 1980), which can increase both cost and risk of infection as
well as physical discomfort to the animal. An alternative is the use of a video-based
remote eye tracker placed close to the animal’s head which allows a combination of
the pupil and/ or one or more Purkinje image to be tracked, this is typically achieved2

by illuminating the eye using an infrared source, the resulting image being capturedby the eye tracker via a camera. Such a technique has been employed with cats
(Körding et al., 2001), mice (Stahl et al., 2000), dogs (Jacobs et al., 2006) andmacaques (Davis et al., 2009). Whilst this technique offers some methodologicalrefinement by eliminating the need to attach apparatus to the eye, owing to the fixednature of the cameras it requires the animal’s body to be restrained and, often, thehead to be fixed, this typically necessitates the surgical insertion of implants into thehead which can be attached to external apparatus. Hence whilst developments inhead-fixation techniques appear to have reduced the incidence of infection andcomplications such as bone necrosis (e.g. Davis et al., 2009), the expense, potential
stress and possibility of harm to the animal associated with surgery cannot becompletely overcome. Not only does fixation of the head and body reduce theopportunities for studying naturalistic behaviour, it also raises concerns over thewelfare of the animals used. All scientists working with animals should show acommitment to the 3 R’s (Russell and Burch, 1959), and so refinement which
improves welfare should be considered important.

The study of the eye movements of freely moving lemurs has been facilitated by theuse of a head-mounted, video-based, eye-tracking system adapted from equipment
designed for human use (Shepherd and Platt, 2006). This system comprised two small
cameras, one recording the visual scene, the other imaging the eye via a reflection
from a dichroic mirror. Such a system, however, has currently not been successfully
used on non-primate species.

3The dog has long been a model laboratory animal but there is a growing interest incanine cognition in many other contexts. Dogs are of particular interest in the studyof social cognition as they are a social species, and therefore likely to be adept atrecognising communicative cues. Their history of artificial selection by man and theiropportunity for enculturation within the human environment (Virányi et al., 2004)
make them a particularly useful model for comparative work with humans.

The study of eye movements and the allocation of visual attention in dogs allows theinvestigation of factors influencing the human-dog relationship, such as the saliencyof human gestures upon dog behaviour (Gácsi et al., 2004; Virányi et al., 2004) andvisual processing biases, providing information on putative cognitive mechanisms
underlying canine vision (Guo et al., 2009; Racca et al., 2010). Hence, studying gaze
patterns has the potential to explore how visual inputs influence a dog’s behaviour aswell as how these inputs may be processed by the visual system. In addition, it
provides a valuable mechanism for studying visual attention itself, both in terms of
how it is deployed and also maintained in dogs, a research area that has received little
attention. The assessment of looking behaviour in dogs has, however, previously
relied largely upon techniques such as measuring the dog’s head and body orientation(Gácsi et al., 2004; Virányi et al., 2004) or change of gaze direction (preferential
looking) (Guo et al., 2009; Racca et al., 2010). Using either of these methodologies,
judgements regarding the allocation of visual attention are restricted in terms ofspatial accuracy. Hence whilst these paradigms are useful for assessing variables suchas whether the dog oriented their attention towards a stimulus, for example a person,
or whether the eyes were attending to the left or right side of a stimulus, they lackspatial accuracy to make more detailed evaluations of the focus of attention.

4

Electrooculograms (EOG) have been used to measure cataplexy in narcoleptic dogs
(Reid et al., 1996, 1998), however, this necessitates the surgical implantation ofelectrodes near the orbit of the eye. The EOG signal is also subject to drift, requiring
regular recalibrations, this is particularly problematic when working with non-verbal
subjects (Aslin and McMurray, 2004). Accuracy of EOG data may be further reduced
by nonlinearities (Dell’Osso and Daroff, 1999). In addition, as the EOG techniquemeasures eye movement within the head it is unsuitable for providing point of regard
information unless head movement is also measured (Duchowski, 2003). Eye
movements in canines have also been studied using infrared reflection tracking
systems in conjunction with body stabilisation and a non-invasive, head-fixation
technique in order to study the effects of a new surgical treatment for infantile
nystagmus syndrome (Dell’Osso et al., 1998, 1999). This methodology has beenfurther advanced by the adaptation of a head-mounted, video-based tracking system,to assess eye movements in dogs with nystagmus (Jacobs et al., 2006). This
necessitated mounting the eye cameras on a fixed frame in front of the dog, the dog’s
body being maintained in a sling with the head manually restrained by anexperimenter. Hence, both of these methodologies still limit the range of naturalistic
behaviours that can be performed by the subject as well as constraining the proportionof the visual scene which can be viewed.

This paper describes the development and assessment of the accuracy of a headmounted, video-based, eye-tracking system for use on dogs. This equipment isintended to provide a more spatially accurate measure of canine looking behaviour
than techniques such as preferential viewing whilst allowing the subject to perform a5far greater behavioural repertoire than permitted by previous canine eye-tracking
methodologies.

2. Methods
2.1 Apparatus
We adapted a VisionTrak head-mounted eye tracker (ISCAN ETL 500, Polhemus,
Vermont, USA) to record gaze patterns from freely moving domestic dogs. The
system has a head-mounted eye and scene imager (consisting of a scene camera, aneye camera, an infrared source and a dichroic mirror) which is connected to a hostworkstation (comprising an RK 826PCI Pupil / Corneal Reflection Tracking
Processor and RK 630PCI Autocalibration System) through a cable 4m in length.
When used on human participants, this robust eye tracker can collect pupil size, eyemovement, and eye point of regard data while allowing complete freedom of head
movement. The system has a sampling rate of 60 Hz and can achieve spatial accuracy
up to 0.3° when used on humans (ISCAN, 2003).

In order to attach a head-mounted eye and scene imager on the dog the apparatus was
mounted on an aluminium head strap (Fig. 1). This provided the head strap withrigidity whilst still allowing it to be lightweight and shaped around the dog’s head.
The strap was attached to the top of a basket muzzle (Baskerville, size 8, The
Company of Animals) using a M10 × 20mm screw with M2 × 10mm screws locatedeither side of it, all secured using nuts. Three screws were used to prevent any lateral
movement of the strap. The heads of the screws, which were located inside themuzzle, were countersunk and concealed behind a leather strap, preventing possible
discomfort or injury to the dog. The head strap extended along the contour of the6

dog’s head, and attached to the muzzle strap at the back of the head by passing
underneath it and folding back on top of it. The muzzle strap was held secure within
this loop using a M4 ×10mm screw and nut positioned immediately in front of it in
the head strap.

Fig. 1. The eye tracking equipment in place on the dog showing scene camera (1) eye
camera (2) head strap (3) mirror support rod (4) mirror clamp (5) dichroic mirror (6)muzzle (7) and mirror frame (8).

Owing to the different head shape of dogs compared to humans it was decided tomount the dichroic mirror in front of the eye rather than below it, as is often the casewith head-mounted, eye-tracking systems designed for human use. In addition, due tothe diversity of head and eye sizes amongst dog breeds it was necessary to be able toadjust the distance of the mirror from the eye to achieve a clear eye image. To
address these issues the mirror was mounted from a square aluminium rod which ran
perpendicular to the dog’s nose: the mirror support rod. Using a frame and clamp
7

device, described below, the mirror could be extended from one of six holes of 2mm
diameter which were drilled horizontally into the mirror support rod at 7mm intervals,
with the first positioned 5mm from the tip of the rod. Lateral movement of the mirror
support rod was inhibited by a second rod, positioned on top of the central nut
attaching the head strap to the muzzle, which projected outwards away from the headstrap. A piece of threaded rod ran vertically through the end of the mirror supportrod, secured by a nut above and below the rod, and located into a hole at the end of
this second rod, providing stabilisation.

In order to obtain a clear eye image from a dog with eyes positioned more laterally
than those of a human it was necessary for the dichroic mirror to move laterally andalso pivot in the horizontal plane, as well as retaining the ability to rotate around thevertical plane, so that it could be positioned parallel to the eye. To facilitate this themirror was held in an aluminium frame 3mm wide and 1mm thick which was shaped
around its outer edge, with three aluminium supports 15mm long and 3mm widelocated equidistantly around the edge of the frame. The supports were attached to theframe using super glue and curved around either side of the mirror to hold it in place
within the frame. Pieces of cushioned adhesive pad were inserted between the
supports and the mirror in order to prevent scratching of the dichroic coating. Both ofthe ends of the strip of metal forming the frame were held in an aluminium block
20mm long, 6mm wide, 6mm deep with a horizontal incision 7mm long and 3mm
wide, the mirror clamp, and secured using a 1M × 13mm screw that passed throughboth the frame and block. This allowed the mirror to be pivoted towards and awayfrom the eye and secured in position by tightening the nut which held the bolt inplace. The mirror clamp was threaded onto a piece of threaded rod of 2mm diameter8

which passed through the 3rd hole in the mirror support rod and was held in place by
two self-locking nuts positioned either side of the mirror support rod, allowing lateral
movement and rotation around the vertical plane.

As the optimal eye image was obtained when the eye camera was parallel to thedichroic mirror, the eye camera was also manufactured to move laterally, pivot aroundthe horizontal plane and rotate around the vertical plane. A square aluminium block
20mm × 20mm and 10mm deep was mounted on the head strap at the highest
position on the dog’s head. In front of this was positioned a second aluminium blockof the same dimensions into which were drilled 2 holes of 5mm diameter, runningparallel through the horizontal section of the block and located 10mm apart. Twoaluminium rods of 5mm diameter were positioned in the holes, the higher of thesepassed into the scene camera casing, the second was 100mm long and incorporated aswivel joint which allowed the final 15mm to pivot around the horizontal plane. A
threaded rod of 2mm diameter and 20mm length was threaded into this end section of
the second rod and passed through a pair of locking nuts which tightened either sideof an aluminium frame, 8mm wide and 1mm thick, shaped around the eye camera tosecure it in place. The position of the two camera rods within the block was secured
by screws which passed into the front of the block, one located 5mm × 5mm from the
top left corner of the block and one 5mm × 5mm from the bottom right corner. Whentightened, these screws exerted sufficient pressure on the rods to clamp them in place,
permitting the eye camera to be moved laterally as well as rotated in the vertical planeand stabilised in the chosen position. The two aluminium blocks supporting thecamera rods were connected via a 1mm threaded rod, 28mm long, mounted into a9section of aluminium rod. The threaded rod passed through the centre of the two
blocks and screwed into the mirror support rod.

2.2 Subject
The subject used for proof of principle was a male Alaskan Malamute, aged two
years.

2.3 Calibration
The calibration procedure required the dog to visually acquire five points in space,
one in each corner of the output captured by the scene camera and one in the centre of
the image. In order to ensure standardisation of the calibration procedure the distance
between the calibration points and the distance between the dog and the calibration
points was fixed. This was achieved using a light metallic cross frame which could be
mounted onto the headgear (Fig. 2). The cross consisted of four aluminium rods,
118mm long and of 6mm diameter each mounted centrally into the sides of a 20mm
× 20mm aluminium block, 8mm deep. An aluminium rod 6mm × 6mm and 235mm
long was centrally mounted into one square face of the block. The cross could then be
mounted to the mirror support rod in front of the cameras via a 6mm × 6mm
aluminium rod 60mm long which was tapered at the end to fit smoothly against theslope of the rod. This was attached to the mirror support rod via a 2M × 23mm and a
2M × 30mm bolt, which ran vertically through both rods. This was connected to ahollow rod 8mm × 8mm and 25mm long, via a 2mm × 20mm section of threaded rod
which passed vertically through both. The calibration device could then be slotted
into this hollow rod and removed once calibration had been achieved. Cardboard
squares 25mm × 25mm were attached to the end of each of the four rods forming the10

cross and in front of the central square. This enabled easy identification of the four
corner points from the scene output monitor and provided the experimenter with clear
targets at which to hold stimuli during the calibration process. This device created afive point calibration system with the four corner calibration points being an equal
distance of 25° away from the central calibration point.

Fig. 2. The calibration cross in place on the head gear showing cross mount (1) andcalibration cross (2).

A stimulus, a treat 10mm x 10mm (2.1 × 2.1º), was held in the centre of each of thefive squares in turn, when the dog was judged to be fixating on the treat in any given
location the calibration point was entered into the computer. Once all five points had
been acquired in this way calibration accuracy could be tested.

11

2.4 Training
In order to habituate the dog to the apparatus it was initially muzzle-trained using
food-based positive reinforcement. Once the muzzle was tolerated for periods up to
30 minutes, the head strap was added and the habituation procedure repeated, thisprocess was repeated twice more for the addition of the cameras and the mirror.
In order to carry out calibration the dog was trained to visually track a treat using eye
movements with minimal movement of the head. This was achieved using positive
reinforcement; an audible click was used to mark the desired behaviour more
precisely. Initially, visual following of a treat was accompanied by large head
movements, however, over approximately 10 training sessions head movements
became minimal and eye movements increased. The same methods were used to train
the dog to maintain fixation on the treat when it was held stationary. The dog was thengradually habituated to the calibration cross being slotted into the front of theheadgear over approximately 5 training sessions, the cross was initially tolerated for
10-20 seconds without behaviours which attempted to remove it, such as pawing at
the device, this was increased to 2-3 minutes during training. Following this the dog
was reinforced for fixating on treats held at the five calibration point locations.

2.5 Data collection
The eye tracker was calibrated as described in section 2.3. The treat was held at eachof the calibration point locations in such a way that the experimenter’s hand and arm
did not enter the space between the five points in order to prevent this acting as a
distraction. When attracting the dog’s attention towards the treat the experimenter
called the dog’s name and pointed to it. If the dog continued not to look at the treat it
was removed from its position and then replaced. For accuracy, only footage with no12

movement or vocalisation was coded. The order in which the treat was held at the fivelocations was randomised between trials in order to prevent learned behaviour
influencing the results. No rewards were given during the trial; once the trial had
ended the dog received the treat. All output from the eye tracker was recorded onto
DVD via a Logik LDVR808 DVD recorder. Nineteen trials were conducted of which
five provided codeable data for all five calibration points.

2.6 Treatment of data
Output from the eye tracker was coded frame by frame at a rate of 30 frames per
second, using a DVD player and a 290mm × 230mm Philips LDH2114\10 video
monitor. For each of the five calibration points the distance was measured between
the centre of the treat when held in position and the centre of the output crosshair in a
direct line. The vertical and horizontal distance between the output crosshair and thestimulus was also measured. The visual angle between the treat and the crosshair was
calculated for each of these measurements on a frame-by-frame basis.

As previous information concerning the use of eye tracking technology on dogs isextremely limited there is a lack of data regarding the features of fixations in thisspecies during visual exploration. Therefore two parameters were examined in detail:first fixation (the first set of data which matched the fixation criteria) and closest
fixation (the dataset matching the criteria in which the output crosshair was closest to
the fixation point). These two sets of data were considered as eye-tracking research inhumans has demonstrated that first fixations on a stimulus tend to be less accurate
than subsequent fixations as they can be subject to overshoot and undershoot (Bötzel
et al., 1993). To qualify as a fixation the output crosshair had to fall within 30mm

13

(5.8º) of the fixation point (centre of the treat) and not move around the fixation pointat an average speed of more than 25°/second for a given duration. The data were
examined using five different minimum durations for this behaviour. These were 67,
100, 133, 167 or 200ms. 30mm (5.8º) was chosen as the maximum distance that theoutput crosshair could fall from the fixation point, since the minimum distance
between any two fixation points on the metal calibration cross was 120mm (25º) (i.e.
distance from centre of the cross to the centre of the cardboard squares positioned at
the end of the arms). Thus the distance between any two fixation points was at least
twice the radius around them (Fig. 3).

Fixationarea

Cardboardsquare30mm
radius

60mm

Aluminiumrod

14

Fig. 3. Scale diagram of calibration equipment and fixation classification areas.
Scale 1:3.

Accuracy was assessed from the mean distance that the centre of the crosshairremained from the centre of the fixation point during a fixation. Data for each of thefive calibration points were averaged across the five trials, providing an overall
accuracy level for each point. Accuracy for first and closest fixations was obtained foreach of the fixation durations; these were then averaged across all durations to
calculate the overall accuracy of the system for first and closest fixations. Horizontal
and vertical accuracy between the crosshair and the fixation point was calculated foreach fixation in the same manner.

2.7 Statistical Analysis
Initial summary descriptive statistics were calculated for accuracy based on mean
deviation from centre of fixation point. Accuracy for first versus closest fixation fordifferent fixation durations was compared using a repeated measures ANOVA. In
order to examine the relationship between accuracy and fixation length for the closest
fixation data, regression analysis using a quadratic equation was used following
inspection of its graphical representation. Evaluation of these results was used todetermine the recommended optimal fixation duration to use in practice.

15

3. Results
The overall accuracy using different fixation duration criteria was around 3° (Table
1).
______________________________________________________
Minimum length

Accuracy (º)

Accuracy (º)

of fixation (ms)first fixation

closest fixation

Mean±SD

Mean±SD______________________________________________________
67

3.76±0.58

2.25±0.47100

3.69±0.73

2.51±0.6133

3.48±0.81

2.6±0.58167

3.62±1.06

2.67±0.62200

3.41±1.09

2.71±0.67

Table 1: Accuracy of crosshair location on video output in relation to predetermined
focal points for fixations defined by varying durations. First fixation = first dataset
matching fixation criteria, closest fixation = dataset matching fixation criteria during
which output crosshair was closest to defined focal point.

The mean level of accuracy for the first fixation across all durations was 3.59° ± 0.07
(radius±SEM). The mean level of accuracy for the closest fixation was 2.55° ± 0.08.
Closest fixation produced a higher level of accuracy (repeated measures ANOVA:
F(1,4) = 55.93 p = 0.002 partial η2 = 0.93).

16

The mean vertical and horizontal accuracy for the first fixations across all durations
were 2.31º ± 0.09 and 2.53 º ± 0.04, respectively. The mean levels of vertical andhorizontal accuracy for the closest fixations were 1.63º ± 0.03 and 1.85º ± 0.07.
Vertical accuracy was greater than horizontal accuracy for the closest fixations(repeated measures ANOVA: F(1,4) = 31.390 p = 0.005 partial η2 = 0.89). No
significant differences were found between vertical and horizontal accuracy for the
first fixations. Horizontal and vertical accuracy across all fixation durations for theclosest durations are shown in Table 2.
____________________________________________________________
Minimum length

VerticalHorizontalof fixation (ms)accuracy (º)

accuracy (º)______________________________________________________
67

1.51±0.33

1.66±0.31100

1.61±0.3

1.74±0.44133

1.65±0.4

1.85±0.33167

1.68±0.38

1.95±0.41200

1.69±0.4

2.03±0.52_______________________________________________________

Table 2: Vertical and horizontal accuracy of crosshair location on video output for
closest fixations defined by varying durations. Values presented in the table are
Mean±SD

Regression analysis using a quadratic equation produced a model with excellent fit(adjusted R squared = 96.4 %) to describe the relationship between accuracy and the17duration used to define a fixation (accuracy = 1.648 + 0.01125fixation – 0.00003
fixation²). Visual inspection of the data, suggests that the point of maximum inflectionof the curve occurs around 100ms (Fig. 4) and that accuracy stabilises beyond 200ms.
Since the apparent accuracy will be affected by both the number of frames sampled aswell as the reliability of the system per se, the optimal fixation duration to use inpractice is based upon consideration of both the relationship between accuracy andfixation duration as well as the point at which this relationship stabilises. On this basis
we suggest that 100ms represents the minimum optimal fixation length to use inpractice, since below this time, the apparently high level of accuracy may be an
artefact of the limited data used in its determination.

Data for the accuracy of individual fixation points using the closest fixation criterion
are given in Table 3.

_________________________________________________________
Calibration point

Accuracy (º) minimum

Accuracy (º) minimum

fixation 100ms

fixation 200ms

Mean±SD

Mean±SD_____________________________________________________________
Top left2.84±1.46

2.84±1.46

18

Top right

2.84±1.74

3.48±2.31

Bottom left

2.00±0.80

2.32±0.89

Bottom right

1.75±0.48

1.77±0.51

Centre3.12±1.19

3.12±1.19_

Table 3: Output accuracy levels for individual calibration points

Fig. 4. Accuracy (º) versus minimum fixation lengths for closest fixations with fitted
line. Increase in degrees indicates reduced accuracy.

It is suggested that the following definition be applied in future when using thisequipment to determine fixations by the dog: a period of at least 100ms duration
during which the output crosshair falls within 6º of a region of interest and does not

19

move around the centre of the point of interest at an average speed of more than25°/second. For calibration purposes it is suggested that the closest fixation be used.

4. Discussion
We have developed and described a reliable head-mounted, eye-tracking system
suitable for use on dogs and a standardised calibration procedure for this equipment.
The accuracy level of 2.25-2.71° achieved using this system is within the bounds ofupper accuracy levels obtained recording the eye movements of other non-humanspecies (Shepherd and Platt, 1996; Guo et al, 2003). We have also provided asuggested definition of a fixation for future research utilising this apparatus on dogs.
Our mean vertical accuracy of 1.63º for closest fixations is very consistent with acanine area centralis which extends ± 1.5º vertically (Jacobs et al., 2006). Greater
discrepancy in horizontal accuracy between the current finding of 1.85 º and areported area centralis extending ± 3º (Jacobs et al., 2006) may reflect variation in the
distribution of retinal ganglion cells that has been noted between dogs with different
nose lengths (McGreevy et al., 2006). In addition, our finding that first fixations wereless accurate than subsequent fixations is consistent with dogs employing similar eye
movements to humans when making saccades towards and fixations upon a stimulus
in which the eye positions itself with greater accuracy during subsequent fixations.
This also offers a possible explanation for our finding of a significant difference
between vertical and horizontal accuracy for closest but not first fixations, as during
the first fixation the stimulus may not be positioned on the area centralis of the retina.

Use of a video-based, eye-tracking system on a non-human species without headrestraint does pose some logistical challenges. In particular, the amount of training20

required to habituate the subject to the equipment and to teach the animal to follow
and fixate on a stimulus with minimal head movements in order to achieve accurate
calibration. However, even though the training may be time consuming, it isrelatively straightforward, being based on simple habituation and routinely used
positive-reward operant procedures. A further potential problem is the introduction oferror into the accuracy of data due to minor movements of the equipment on the dog’shead during testing. For this reason it is suggested that output accuracy is assessed
immediately following calibration, using the method described in section 2.5, andagain at the end of each testing block. Restricting the length of testing sessions totime periods for which the dog will tolerate the equipment without becoming restless
would also help to overcome this issue. As calibration is performed with the dog
facing the experimenter the direction of gaze can also be visually assessed as thecalibration points are entered into the system.

The current study utilised a narrow-view scene camera lens intended for studying
close-range dog-dog and dog-human interactions, adaptations to the equipment andcalibration protocol may be necessary when the wide-view scene camera is used forstudying dog’s gaze behaviour towards visual stimuli at greater distances. As theviewing distance of our calibration grid (approximately 30cm) is very close to dogs’
optimal accommodation range (33–50cm; Miller and Murphy, 1995), it is unlikely ourcalibration distance will induce near point of convergence in dogs. Furthermore, we
use dogs with reasonable binocular vision in our research as this reduces thepossibility of convergence, due to the need to maintain binocular vision, during thecalibration procedure. In addition, this measure will facilitate the comparison of
acquired data with that from humans and non-human primates.

21

Use of a monocular eye tracker can have limitations and present challenges.
Information regarding the intersection of gaze angles from both eyes is not available
to facilitate the identification of targets at depths different to that for which theequipment has been calibrated. Care must also be taken when selecting appropriate
subjects, for example, achiasmatic dogs have been shown to display monocularsaccades (Dell’Osso and Williams, 1995) and hence representative eye movement
data could not be obtained from such animals using a monocular device. In addition,
dogs in general may show looser yoking of the two eyes than is found in humans andnon-human primates (Dell’Osso and Williams, 1995). If this is the case then thecurrent methodology of conducting a binocular calibration, with both eyes viewing
the stimuli, may result in a less accurate output than monocular calibration, in which
only the eye being tracked would view the calibration stimulus, ensuring that stimuli
are fixated by the correct eye. Binocular calibration was employed in the current
study as it can be performed more quickly, is less intrusive and requires less training
than monocular calibration. In addition, in some circumstances, such as our intention
to study gaze behaviour in naturalistic settings, dogs use both eyes to view scenes.
Therefore, given that monocular calibration with a monocular eye-tracker would only
allow one eye to view the calibration points, binocular calibration may be more
ecologically valid and more comparable with data obtained from a dog viewing
binocularly. The protocol of binocular calibration with a monocular eye-tracking
system is not uncommon in studies of fixational and scene viewing gaze behaviour inhumans and non-human primates (e.g. Guo et al., 2003; Guo, 2007; Steckenfinger andGhazanfar, 2009; Shepherd et al., 2010). Previous studies have revealed that during asaccade, the eyes may initially diverge, but convergence occurs in the later part of the22

saccade, continuing into the following fixation period (e.g. Collewijn et al., 1995),
suggesting that maximum disparity between the eyes occurs during saccades withincreased convergence during fixations. Indeed, it has been suggested that for tasks
that do not require far distance or depth perception, such as reading, monocular andbinocular calibration produce comparable results in human adults (Nuthmann andKliegl, 2009). Given the relative infancy of canine eye-tracking, particularly using
monocular equipment, there is a lack of information regarding the effects ofmonocular and binocular calibration; further research in this area is necessary to
inform future calibration protocols.

Despite these caveats the monocular system employed here provides a number ofbenefits for conducting behavioural research with dogs. It is unobtrusive; the only
equipment placed in front of the eye is the transparent mirror and the lack of restraint
required may permit longer recording sessions than would be tolerated by a restrained
subject. The equipment is relatively inexpensive and can be easily operated by asingle experimenter. In addition, calibration can be conducted quickly, which ishighly beneficial when working with non-verbal subjects. Monocular systems havebeen commonly used to study natural vision in humans navigating through theenvironment (Cheong et al., 2008) and when looking at more detailed stimuli such as
faces (Kleinhans et al., 2008), as well as in non-human animals (Shepherd and Platt,2006).

The five calibration point locations, the centre and four corners of the scene output,
were chosen to assess the accuracy of the system as these positions are typically used
for rapid calibration in head-mounted eye trackers. Use of these positions may be

23

more susceptible to cross-talk errors between the horizontal and vertical axes
compared to points actually located on the two axes (i.e. left, right, top and bottom).
However, overall, the use of these corner points is likely to allow a more naturalistic
account of eye movements by considering non-linear interactions between thehorizontal and vertical planes.

Our system appears to provide a greater level of accuracy than the head-mounted
ISCAN system used with macaques by Shepard and Platt (2006), which achieved an
accuracy of 5-10°. This may reflect the more standardised calibration procedure
employed in the current study. The calibration cross enabled standardisation of the
distance between both the calibration points and the subject and the distance between
the points themselves, whilst Shepherd and Platt (2006) calibrated their equipmentwith a trainer holding treats entering the visual scene and standing at the fivecalibration locations, permitting less standardisation of distances. Whilst the current
technique provides less spatial accuracy than that provided by the scleral search coiltechnique employed by Guo et al. (2003) in primates, which achieved an accuracy
level under 1°, the current technique permits investigation in more naturalistic
settings, allowing movement of both head and body, as well as avoiding the surgery
required to implant a coil into the eye of a non-human species. By allowing a greater
behavioural repertoire including head movement, a larger proportion of the visual
scene can be scanned by a subject wearing the current device compared to that used inother techniques such as that employed by Jacobs et al (2006), allowing a moreenriched view of visual attention. Hence, whilst not suitable for recording very
detailed eye movement information, it is hoped that this equipment will permit a24

naturalistic method for studying the allocation of dogs’ visual attention in nature
vision.

In future, the authors hope to use this system to study dogs’ visual attention in avariety of naturalistic settings, such as in social interactions both with members of
their own species and humans. It also has the potential to provide a measure of visualacuity both between dog breeds and in specific individuals. As allocation of visual
attention can be used to assess cognitive function (Crutcher et al., 2009), the system
could also be applied to the identification of cognitive changes and degeneration.

In conclusion, this system provides a non-invasive method of assessing dogs’ looking
behaviour without restraint with a higher level of spatial accuracy than previously
available for this species, and has potential application in a wide variety of researchsettings.

Acknowledgements
The authors wish to thank David Williams for his help and guidance with manufactureof the equipment, Paul Coster, George Rodis and Andrew Sherman for their technicalsupport and Petronella Nilsson for her assistance during the training process.

25

References:

Aslin RN, McMurray B. Automated corneal-reflection eye tracking in infancy:
methodological developments and applications to cognition. Infancy, 2004;6:155-63.
Bötzel K, Rottach K, Büttner U. Normal and pathological saccadic dysmetria. Brain,
1993;116:337-53.
Cheong AMY, Geruschat DR, Congdon N. Traffic gap judgment in people withsignificant peripheral field loss. Optometry Vision Sci., 2008; 85:26-36.
Collewijn H, Erkelens CJ, Steinman RM. Voluntary binocular gaze-shifts in the planeof regard: dynamics of version and vergence. Vision Res., 1995;35:3335-58.
Crutcher MD, Calhoun-Haney R, Manzanares CM, Lah JJ, Zola SM. Eye tracking
during a visual paired comparison task as a predictor of early dementia. Am J
Alzheimer’s Dis., 2009;24:258-66
Davis TS, Torab K, House P, Greger B. A minimally invasive approach to long termhead fixation in behaving nonhuman primates. J Neurosci Meth., 2009;181:106-10.
Dell'osso LF, Daroff RB. Eye movement characteristics and recording techniques. In
Glaser JS, editor. Neuro-Ophthalmology. Lippincott, Williams & Wilkins:
Philadelphia, 1999:327-43.
Dell’Osso LF, Hertle RW, Williams RW, Jacobs JB. A new surgery for congenital
nystagmus: effects of tenotomy on an achiasmatic canine and the role of extraocular
proprioception. JAAPOS, 1999;3:166-82.
Dell’Osso LF, Williams RW. Ocular motor abnormalities in achiasmatic mutant
Belgian sheepdogs: unyoked eye movements in a mammal. Vision Res.,
1995;35:109-16.

26

Dell’Osso LF, Williams RW, Jacobs JB, Erchul DM. The congenital and seesaw
nystagmus in the prototypical achiasma of canines: comparison to the humanachiasmatic prototype. Vision Res., 1998;38:1629-41.
Duchowski AT. Eye Tracking Methodology: Theory and Practice, first ed. Springer:
Berlin, 2003: 56.
Gácsi M, Miklόsi A, Varga O, Topál J, Csányi V. Are readers of our face readers of
our minds? Dogs (Canis familiaris) show situation dependent recognition of humans’attention. Anim Cogn., 2004;7:144-53
Guo K. Initial fixation placement in face images is driven by top-down guidance.
Exp Brain Res., 2007;181:673-77.
Guo K, Meints K, Hall C, Hall S, Mills D. Left gaze bias in humans, rhesus monkeys
and domestic dogs. Anim Cogn., 2009;12:409-18.
Guo K, Robertson RG, Mahmoodi S, Tadmor Y, Young MP. How do monkeys view
faces? – A study of eye movements. Exp Brain Res., 2003;150: 363-74.
Henderson JM. Human gaze control during real world scene perception. Trends
Cogn Sci., 2003;7:498-504.
ISCAN Inc. ISCAN ETL-500 Operating Instructions. ISCAN Inc.: Woburn, 2003:3.
Jacobs JB, Dell’Osso LF, Hertle RW, Acland GM, Bennett J. Eye movement
recordings as an effectiveness indicator of gene therapy in RPE65 deficient canines:
implications for the ocular motor system. Invest Opthalmol Vis Sci., 2006;47:286575.
Judge SJ, Richmond BJ, Chu FC. Implantation of magnetic search coils formeasurement of eye position: an improved method. Vision Res., 1980;20:535–38.

27

Kleinhans NM, Richards T, Stirling L, Stegbauer KC, Mahurin R, Johnson LC,
Greenson J, Dawson G, Aylward E. Abnormal functional connectivity in autism
spectrum disorders during face processing. Brain, 2008; 131:1000-12.
Körding KP, Kayser C, Betsch BY, König P. Non contact eye tracking on cats. J
Neurosci Meth., 2001;110:103-11.
Land MF. Eye movements and the control of actions in everyday life. Prog Retin
Eye Res., 2006;25:296-334.
McGreevy PD, Grassi TD, Harman AM. A strong correlation exists between the
distribution of retinal ganglion cells and nose length in the dog. Brain Behav Evolut.,
2004;63:13-22.
Miller PE, Murphy CJ. Vision in Dogs. J Am Vet Med Assoc., 1995;207:1623-34.
Nuthmann A, Kliegl R. An examination of binocular reading fixations based on
sentence corpus data. J Vision, 2009;9:1-28.
Racca A, Amadei E, Ligout S, Guo K, Meints K, Mills D. Discrimination of humanand dog faces and inversion responses in domestic dogs (Canis familiaris). Anim
Cogn., 2010;13:525-33.
Rayner K. Eye movements in reading and information processing. Psychol Bull.,
1998;124:372-422.
Reid MS, Nishino S, Tafti M, Siegel JM, Dement WC, Mignot E.
Neuropharmacological characterisation of basal forebrain cholinergic stimulated
cataplexy in narcoleptic canines. Exp Neurol., 1998;151:89-104.
Reid MS, Tafti M, Nishino S, Sampathkumaran R, Siegel JM, Mignot E. Local
administration of dopaminergic drugs into the ventral tegmental area modulates
cataplexy in the narcoleptic canine. Brain Res., 1996;733:83-100.

28

Russell WMS, Burch RL. The Principles of Humane Experimental Technique, firsted. Methuen: London, 1959.
Shepherd SV, Platt ML. Noninvasive telemetric gaze tracking in freely movingsocially housed prosimian primates. Methods, 2006;38:185-94.
Shepherd SV, Steckenfinger SA, Hasson U, Ghazanfar AA. Human-monkey gazecorrelations reveal convergent and divergent patterns of movie viewing. Curr Biol.,
2010;20:649-56.
Stahl JS, van Alphen AM, De Zeeuw CI. A comparison of video and magnetic searchcoil recordings of mouse eye movements. J Neurosci Meth., 2000;99:101-10.
Steckenfinger SA, Ghazanfar AA. Monkey visual behavior falls into the uncanny
valley. P Natl Acad Sci USA, 2009;106:18362-466.
Virányi Z, Topál J, Gácsi M, Miklόsi A, Csányi V. Dogs respond appropriately tocues of humans’ attentional focus. Behav Process., 2004;66:161-72.

29