WavelengthCombiners
A powerful and highly integrated solution
for particle analysis, flow cytometry andmicroscopy applications.

Up to 4
diode lasers

Independent TTL
modulation for all
wavelengths+12 VDCpower supply

Free space
SM fiber
MM fiber
PM fiberAdvantages− 4 slots for detectors/emitters− Free-space or multi-mode fiber output− Color mixing− Fast warm-up time (bi-directional TEC)
− Compatible with MatchBox accessories
A dedicated Break-out-Box can be purchased separatelly. It provides PD-type power supply support, fancontrol , an interlock. and inputs for TTL modulation.

Thermoelectric

3 diode and
1 DPSS/OPS
lasersWorld’s smallest
form factor

Vast beam
shaping options

Extrememechanicalstability

Low parasitic
reflections

Applications
Flow cytometry

Particle analysis

Sorting

Microscopy

Illumination

Spectroscopy

Ophthalmology

Part numbers of combiners
Based on the item code structure there are thousands ofdifferent configurations, which are theoretically possible.
You are welcome to contact our sales staff in order to get a
preliminary evaluation if your target configuration is potentially
feasible in the MatchBox combiner platform.

Output types
Integrated Optics uses proprietary micro- optics
assembly technique to assemble tiny opticalcomponents inside the MatchBox Combiner.
Fiber coupling is permanently carried out inside thebox, thus alignment-free operation is ensured.

MatchBox

Platform

ARA

s are

Multi-wavelength laser specifications
Part No.

Wavelength
set, nmOutput power(free-space),mW

Output power(MM fiber),mW

Output power(SM/PM fiber),mW

40A-45A-48A-52A-1...

405
450
488
520

120
70
40100

10050
40
70

30
2020300,50,81140A-45A-48A-64A-1...

405
45052063812070100
1301005070100

302030300,50,810,740A-48A-52A-64A-1...

405
488
520
638

120
40100
1301003070100

302030300,51
10,740A-48A-64A-78A-1...

405
488
638
785

120
40130120

100
30
100
100

30203030

40A-52A-64A-78A-1...

405520
638
785

120
100
130120

100
90
100
100

30
30
30
30

0,5
10,70,240A-48A-52B-64A-2...

405
488
532
638

120
40100
1301004090
100252025250,510,10,740A-48A-56B-64A-2...

405488561638

12040501301004040100252020250,510,20,7Power stability,% (RMS, 8 hrs)*

Free-space <0.2
MM fiber <0.5SM fiber <1PM fiber <1Spectralline-width,nm*

0,5
10,70,2* Typical performance (full specification available at https://integratedoptics.com/products/wavelength-combiners)
Custom: other wavelengths on request: 505 nm, 660 nm, 830 nm, 850 nm, 1064 nm, 1470nm, 1550 nm
NOTE: Product specifications are subject to change without prior notice to improve reliability, function or design or otherwise.

Following output types are available/
customizable:− Collimated Free-space− Line-focussed− MM fiber-coupled− SM fiber-coupled− PM fiber-coupled− SMA portOther parameters of combiners
Beam properties:− Beam diameter at aperture (1/e2): <2 mm− Beam divergence (full angle): <1.5 mrad− Beam pointing stability: <5 μrad/C°

Operation mode:− Automatic Current Control (ACC)− TTL modulation up to 10 MHz in ACC mode. Each laser diode can be modulated independently.

Fiber specs:− Default connector for SM/PM fiber is FC/PC− Default connector for MM fiber is SMA− Standard length of a fiber is 1 m to 1.2 m− Polarization rotation (PM fiber): less than 5 degree

Physical properties:− Control interface type: UART serial bus, convertible to USB or RS232 using accessories− External power supply requirement: 1.5 A and +12 VDC− Dimensions (L-W-H): 50 x 30 x 18 mm (excluding pins and output window)− Beam height from the base: 10.4 mm (+/- 0.3 mm)− Heatsink requirement: <0.5 °C/W
− Optimum heatsink temperature (non-condensing): +15...+30 °C− Max. heatsink temperature 40 °C− Internal temperature stabilization: TEC− Overheat protection: Yes− Warranty: 14 months, or 10000 hours, whichever occurs first. Operational time calculation isbased on an internal EPROM counter

Compatibility:− RoHS− Electromagnetic Compatibility (EMC) Directive 2004/108/EC− General Product Safety Directive (GPSD) 2001/95/EC− IEC60825-1:2014 (compliant only using additional accessories)
www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067 FAX:048-871-0068

The Pinout
The bottom row comprises Vcc and GNDpins, that are used for 12 VDC power
supply; Tx and Rx pins are for UART
communication; TTL pin is universallyprogrammable and is set to fan-control
mode as a default.
The row is used for TTL modulation ofeach installed laser diode.

e-mail:voc@phototechnica.co.jp