thsince 1944

www.nichiryo.co.jp/en/2024-

Distributor

Tokyo Oﬃce
PMO Kandaiwamoto-cho II 4F, 2-2-4 Iwamoto-cho
Chiyoda-ku, Tokyo 101-0032 Japan.
TEL. +81-3-5829-4539 FAX. +81-3-5829-4504

0208192024

2024.08. Updated

NICHIRYO

C ONTE NTSPR O D U C T LI N E U P

Future Lifescience Partner
We will continue to contribute in the endeavor

05

Nichipet Air

to make a better society through supporting07

Nichipet Air -Multi Channel- NEW

our customers in their business solutions.

09

Nichipet Premium

10

Nichipet Premium S-PAC

11

Nichipet EX PlusⅡ

13

Nichipet EXⅡ

14

Nichipet EXⅡ S-PAC

15

NICHIRYO Le

Today, the Nichiryo liquid handling technology supports customers worldwide in

16

Nichipet ECO

various fields from life science to clinical laboratories.

17

Nichipet FⅡ&VⅡ / MODEL 3100

18

Premium Tip

19

NichimateAqua™ series NEW

20

Pipette Mate NEO

21

Nichimate Stepper

22

MODEL 2100DG / DISPET

23

DISPET EXⅡ / DISPET MiniACCESSORY

24

ACCESSORYCUSTOM MADE & OEM

25

Custom-made Pipettes / NSS-1000ⅡBasicEver since establishment in 1944, Nichiryo Co.,LTD.
has been contributing to society as a liquid handling device
manufacturer through supporting customer prosperity.

PIPETTE

We will pursue further global operations to bond a profound partnership withcustomers by offering proactive solutions with "Challenge,
Creation and Cooperation".
President Yoshinori Ito▲

Corporate Identity

DISPENSER

Mutual prosperity with environment, life and sustainable development.
Nichiryo as a faithful partner, promise to provide reliable products,
service and solutions to all researchers.

CONCEPT「i」like human being means Nichiryo always connects with customers in their trust. It also
looks like a flowing movement and a broad expanse. We will keep moving toward the future.
Adopting green color shows we always devote ourselves to being an environment conscious company.
To ensure safe and proper use of our products and to avoid harm to people and property, please make sure to carefully read andfully understand the “Safety Precautions” (especially the “Cautions” section) before use. If anything is unclear, please contact us.

▲

Guidelines for Action

For accidents or harm caused by misuse and altering of the products without Nichiryo’s permission, we will not be held responsible.
The speciﬁcations, usage and designs are as of August 2024.
The values of accuracy and precision are made with distilled water at room temperature 22°C-23°C.

01

page.
www.nichiryo.co.jp/en/

1

Challenge

Always making efforts to achieve the high goal.

2

Creation

Thinking differently. Creating innovative products.

3

Cooperation

Being always aware of building a cohesive team.

Speciﬁcations of instruments and optional accessories as well as the contents of the accessory sets are subject to change
without notice.
The products in this catalogue have been photographed without warning labels.
Windows © is a registered trademark of American Microsoft Company. Pentium® is a registered trademark of American Intel Company.

page.
www.nichiryo.co.jp/en/

02

Customer Support
Quality Assurance System

Calibration and Maintenance Service

Quality Assurance System and Product Management

In order to maintain products in the best condition, we provide calibration
service at JCSS accredited laboratory corresponding to international MRA.
Our calibration service center uses ISO/IEC 17025（JIS Q
17025）as accreditation criteria and is accredited by
JCSS which is operating an accreditation scheme inaccordance with ISO/IEC 17025.The accreditation body
（IA Japan）
operating JCSS has signed Mutual Recognition
Arrangement（MRA）of Asia Pacific Laborator y
Accreditation Cooperation（APLAC）and International
Laboratory Accreditation Cooperation（ILAC）.
Our calibration service center is a JCSS accredited
laboratory corresponding to international MRA.The
accreditation number of our calibration service center is
JCSS0280.

■Products which can be calibrated with JCSS,accreditation criteria.

・Nichipet Series ・NICHRYO Le■Certificate of JCSS and MRA■Ex. JCSS calibration certificate

Based on dispenser technology that has been cultivated since our establishment in1944, the history of Nichiryo's pipettes started from the first micropipette "Justor"
launched in 1975, and now it keeps evolving further as "Nichipet series" underISO9001-based quality assurance system in addition to considerable technological and sales experiences.
Nichiryo is certified according to environmental management system standard ISO14001 of International
Organization for Standardization.
N i c h i r y o i s ke e n l y a wa r e t h at t h e g l o b a l
e nv i ro n m e nt p rote cti o n a n d th e p o l l uti o nprevention are the most important tasks throughall business activities and incidental services, andwe fulfill our responsibilities to the society andregions by environmental protection activities.

5-year Warranty7(675(3257
7HVWLQJSURFHGXUHVFRPSO\ZLWK,62VWDQGDUGV
-,6.
,VVXH1R
3LSHWWHW\SH
6HULDO1R
7LS1R
9ROXPHUDQJH
3URWRFRO1R

20*****-*****

1LFKLSHW3UHPLXP
(
*********
%07/*   XO

5RRPWHPSHUDWXUH &:DWHUWHPSHUDWXUH &$LUSUHVVXUH
K3D&RUUHFWLRQYDOXH= XOPJ
5HODWLYHKXPLGLW\
%DVLVRIDGMXVWPHQW ([ WRGHOLYHU\
7HVW
7HVWYROXPHXO
1RRI7HVW









7HVWUHVXOWV0HDQYDOXHXO/LPLWYDOXH
6\VWHPDWLFHUURU HV  
3DVVHG
5DQGRPHUURU &9  
3DVVHG
7HVW
7HVWYROXPHXO





0HDQYDOXHXO/LPLWYDOXH
6\VWHPDWLFHUURU HV  
5DQGRPHUURU &9  

1RRI7HVW




7HVWUHVXOWV
3DVVHG
3DVVHG

7HVW
7HVWYROXPHXO






0HDQYDOXHXO/LPLWYDOXH
6\VWHPDWLFHUURU HV  
5DQGRPHUURU &9  

1RRI7HVW




7HVWUHVXOWV
3DVVHG
3DVVHG%DODQFH
6DUWRULXV06(3',DXWRGRRU
2QOLQHPHDVXUHPHQW
MSE225P%DODQFH1R

A-01
7KHUPRPHWHU
+3$7KHUPRPHWHU1R
:DWHUWHPSHUDWXUHPHWHU1R70
Ｔ.Nichiryo,GHQWLILFDWLRQRIRSHUDWRU
5.XZDEDUD
7HVW'DWH

2022/01/01'DWHSULQWHG

6LJQDWXUH 
2022/01/01

Nichiryo Co., Ltd.

1. Quality Assurance System

1. The 5-year warranty is only valid when the 5-year warranty registration form iscompleted and returned to Nichiryo.

Nichiryo is accredited with quality management system standard ISO 9001 (JIS Q 9001:2000)
certification, and responsibly execute manufacturing and quality control of our products. We maintain

2. The 5-year warranty does not cover defects caused by excessive wear and tear

our quality assurance system to the high level for our products and services that satisfies users.

or damage due to shipping, accident, abuse, misuse, or usage not in accordancewith instruction manual, or damage caused by liquid permeation in the nozzle cone.

2. Product Standard and Performance Test

3. If you have any questions, please contact local distributor or Nichiryo. Following

Nichiryo a member of the drafting committee of Japan Industrial

is contact information for Nichiryo, Japan.
e-mail : i-sales@nichiryo.co.jp＊Above conditions are subject to change without any notice.

Standards JIS K 0970 (Piston operated micro-volumetric apparatus)
for the standardization of pushbutton type pipettes enacted in 1989,
Nichiryo develops pipettes based on this standard. In addition, theperformance test by international standard ISO8655 (Piston operated
volumetric apparatus) is executed by the customer's demand.

You can visit own member page to see Dream Bank point summary, to add your point, to redeem Nichiryo
items and to receive other various services. In addition, you can receive member campaign information and3. Traceability Management
N ichir yo guarantees the accuracy per formance of products

other special message from Nichiryo.

systematically with calibration traceability to clarify the accuracy of the

Points redemption You can get free items from wide range of selections by redeeming Dream Bank points.

measured value of Nichiryo's reference standard apparatus, such as
the balances, weights, and thermometers, to agree with the national

32

point

64
point

26

point

40

point

20

point

32

point

And
more!!

measurement standard. Nichiryo manages pipettes by the serial
number, and to facilitate the traceability of them, we have constructed
the consistent record system from manufacturing to repair in ourcompany.

Register now! ▶ https://www.nichiryo.co.jp/en/dreambank/
03

page.
www.nichiryo.co.jp/en/

page.
www.nichiryo.co.jp/en/04

Nichipet Air

P I P E T TE

■ Product lineup（8 colors）

Among the worldʼs lightest Micro Pipettes

New evolution for world- class light weight,
accuracy, and precision.
00-NAR-2
0.1〜2µL

00-NAR-20
2〜20µL

00-NAR-10
0.5〜10µL

00-NAR-100
10〜100µL

00-NAR-200
20〜200µL

00-NAR-1000
100〜1000µL

00-NAR-5000
500〜5000µL

00-NAR-10000
1000〜10000µL

Aspiration&
Discharge−70%
Volumesetting■ Stroke loading comparison with competitors（1000µL）
Vo l u m e a d j u s t m e n t b y t h e
exclusive jig is only have to set
the volume into the scale marks.

Up to 70% reduction in stroke
load pressure compared withconventional models.

Vo l u m e s et ti n g rot ato r i s
in d ep en d ent fro m th e p ush
button.
Avoiding the volume shift during
pipetting.

200

4.5

Competitor A

4.0

Competitor B

2.0

Nichipet Air

0

5

10

15

Stroke（㎜）

Keeps the same durability as
previous model and reduce theweight as 56% at maximum.
Operability is improved withcompact design. Approximate
70g. Fully autoclavable.

【Parts List】

00-NAR-10（● Pink）
00-NAR-20（● Cream）■In order to reduce fatigue during long-term use, significant weight reduction and stroke load reduction have been achieved.
■The entire body is autoclaved (121°C, 20 minutes) with possible durability and the highest level of accuracy and repeatability.

05

page.
www.nichiryo.co.jp/en/

75
50

0

25

NichipetAir

Nichipet
EXⅡ

Competitor

A

Competitor

B

Competitor

C

Digital Setting

Volume range(µL)
0.1 〜 2
0.5 〜 102 〜 2010 〜 100
20 〜 200
100 〜 1000
500 〜 5000
1000 〜 10000

Increments(µL)

0.002
0.01
0.02
0.1
0.2
1
5
10

Usable tips

BMT2-UT/UTWR/UTWRB、FUT/FUTRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB
BMT2-X/XR/XRB
BMT2-ZS/ZSR/ZSRB *Accuracy and Precision（● Gray）

The pipette volume (Cat. No.) is
recognized in one glance byvolume notation of 3 positions.

100.5g

89.5g

100*Exclusive tip for 00-NAR-10000. *See p.18 for tip selection.

00-NAR-2

In order to discharge the set
dispensing amount, maximum
volume range of 2μL , 10μL and20μL have high-performanceblow-out function.

20

Volume Setting

00-NAR-2（● Gray）
00-NAR-10（● Pink）
00-NAR-20（● Cream）
00-NAR-100（● Orange）
00-NAR-200（● Yellow）
00-NAR-1000（● Blue）
00-NAR-5000（● Green）
00-NAR-10000（● White）

Cat. No＊This image is 20μL

116g

Specifications
Cat. No

One-touch removable and canbe used for a thin test-tube.

125

25

0.5

Removable witheasy rotating︵ ︶Weightg︵ kg︶
Loading

2.5

1.0

The volume can be checked
during pipetting.

146g

150

Nichipet EXⅡ

3.0

0.0

191g

175

3.5

1.5

T h e s h a p e to ﬁ t t h e h a n d s
based on the ergonomics.

■ Weight comparison with competitors（10000µL）

00-NAR-100（● Orange）

Volume(µL)

0.2
1
2
1
5
10
2
10
20
10
50
100

Accuracy（%）± 12.0 *
± 2.5
± 1.4
± 2.5
± 1.5
± 1.0
± 3.0
± 1.2
± 1.0
± 3.0
± 1.0
± 0.8

Precision（%）≦ 6.0 *
≦ 1.5
≦ 0.7
≦ 1.8
≦ 0.8
≦ 0.4
≦ 1.5
≦ 0.6
≦ 0.3
≦ 1.0
≦ 0.3
≦ 0.2

Cat. No

00-NAR-200（● Yellow）
00-NAR-1000（● Blue）
00-NAR-5000（● Green）
00-NAR-10000（● White）

Volume(µL)

20
100
200
100
500
1000
500
2500
5000
1000
5000
10000

Accuracy（%）
± 1.8
± 1.0
± 0.6
± 1.0
± 0.8
± 0.6
± 2.0
± 0.7
± 0.5
± 2.0
± 0.8
± 0.5

Precision（%）
≦ 0.7
≦ 0.3
≦ 0.2
≦ 0.6
≦ 0.2
≦ 0.2
≦ 0.6
≦ 0.25≦ 0.15≦ 0.6≦ 0.2≦ 0.15*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NAR-2 below 0.2μL and NAR-10 below 1μL depend on the sampling skill and the sampling condition where pipettes are used.

＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

06

Nichipet Air -Multi Channel-

PI PE TTE

■ Product lineup（4 colors）

Among the worldʼs lightest Micro Pipettes-Multi ChannelMulti channel model of 'Nichipet Air'
among the worldʼs lightest pipettewith world class Accuracy and Precision.
Aspiration&
Discharge−50%

00-NAR-100M8
10~100µL

00-NAR-10M8
0.5~10µL

00-NAR-200M8 00-NAR-300M820~200µL
30~300µL

00-NAR-10M12
0.5~10µL

00-NAR-100M12
10~100µL

00-NAR-200M12
20~200µL

00-NAR-300M1230~300µL

Volumesetting

Vo l u m e a d j u s t m e n t b y t h e
exclusive jig is only have to set
the volume into the scale marks.

Up to 50% reduction in stroke
load pressure compared withconventional models.

Vo l u m e s et ti n g rot ato r i s
in d ep en d ent fro m th e p ush
button.
Avoiding the volume shift during
pipetting.

■ Stroke loading comparison in ʻ Nichipet（
ʼ 200 μ L）
3.5

■ Weight comparison in ʻ Nichipet ʼ (100 μ L)
250

NPM-8LP
NPM-12LP

3.0

NAR-200M8
NAR-200M12

2.5

197.1g200︵ ︶Weightg︵ kg︶
Loading

2.0

1.5

239.1g

NPM
NAR

191.8g156.3g

150

100

1.0
50
0.5

The shape to ﬁt the hands
based on the ergonomics.

The volume can be checked
during pipetting.

0.0

0

2

4

6

8

10

12

Stroke
（㎜）

14

16

18

0

20

8ch 100μL

12ch 100μL

Specifications
Cat. No

Keeps the same durability as
previous model and reduce theweight as 20% at maximum.
Operability is improved withcompact design. Fully autoclavable.

The pipette volume (Cat. No.) is
recognized in one glance byvolume notation of 4 positions.

Channel00-NAR-10M8（●Pink）
00-NAR-100M8（●Orange）
00-NAR-200M8（●Yellow）
00-NAR-300M8（●White）
00-NAR-10M12（●Pink）
00-NAR-100M12（●Orange）
00-NAR-200M12（●Yellow）
00-NAR-300M12（●White）

Volume Setting

8
DigitalSetting
12

Volume range(µL)
0.5 – 10
10 – 100
20 – 200
30 – 300
0.5 – 10
10 – 100
20 – 200
30 – 300

Increments(µL)

0.01
0.1
0.2
1
0.01
0.1
0.2
1

Usable tips

BMT2-UT / UTWR / UTWRB 、FUT / FUTRB
BMT2-SG / SGR / SGRB 、FSE / FSERB
BMT2-SG / SGR / SGRB 、FAG / FAGRB
BMT2-K/KR/KRB
BMT2-UT / UTWR / UTWRB 、FUT / FUTRB
BMT2-SG / SGR / SGRB 、FSE / FSERB
BMT2-SG / SGR / SGRB 、FAG / FAGRB
BMT2-K/KR/KRB*See p.18 for tip selection.
【Parts List】

objectAccuracy and Precision
Push
down

＊This image is NAR-10M8■ In order to reduce fatigue during long-term use, significant weight reduction

and stroke load reduction have been achieved.
■The entire body is autoclaved (121°C, 20 minutes) with possible durability

and the highest level of accuracy and repeatability.

07

page.
www.nichiryo.co.jp/en/

In order to discharge the set
dispensing amount,maximum volume range of 10μ
L have high-performance blowout function.

This function improves attachmentof the tips from the rack.

Cat. No
10μL（00-NAR-10M8）（00-NAR-10M12）
100μL（00-NAR-100M8）（00-NAR-100M12）

Volume(µL)
1
5
10
10
50
100

Accuracy(%)
±8.0＊
±4.0
±2.0
±3.0
±1.0
±0.8

Precision(%)
≦5.0＊
≦2.0
≦1.0
≦2.0
≦0.8
≦0.3

Cat. No
200μL（00-NAR-200M8）（00-NAR-200M12）
300μL（00-NAR-300M8）（00-NAR-300M12）

Volume(µL)
20
100
200
30
150
300

Accuracy(%)
±3.0
±1.0
±0.8
±3.0
±1.0
±0.8

Precision(%)
≦1.5
≦0.7
≦0.3
≦1.0
≦0.5
≦0.3

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NAR-10M8 and 10M12 below 1μL depend on the sampling skill and the sampling condition where pipettes are used.

＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

08

Nichipet Premium

P I P E T TE

Specifications

Autoclavable Micro Pipette

Cat. No
00-NPP-2
00-NPP-1000-NPP-20
00-NPP-100
00-NPP-200
00-NPP-1000
00-NPP-5000
00-NPP-10000

The 5-year warranty is proof of high durability.

Volume Setting

Volume range(µL)
0.1 – 2
0.5 – 102 – 20
10 – 100
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Digital Setting

Increments(µL)
0.002
0.01
0.02
0.1
0.2
1
10
10

Usable tips
BMT2-UT/UTWR/UTWRB、FUT/FUTRB
BMT2-SSW/SSWR/SSWRB 、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB
BMT2-X/XR/XRB
BMT2-Z/ZR/ZRB*See p.18 for tip selection.

Accuracy and Precision
Cat. No
00-NPP-2

00-NPP-1000-NPP-20

00-NPP-100
Enlarged and easy to read counter
display for volume setting.

New material used for 4 notchelock improves its strength,
which minimizes
the lever shift fromthe original position.

Accuracy（%）±12.0*
±5.0
±3.0
±4.0*
±1.0
±1.0
±5.0
±1.0
±1.0
±2.0
±1.0
±0.8

Volume(µL)
0.2
1
2
1
5
10
2
10
20
10
50
100

Precision
（%）
≦6.0*
≦2.5
≦1.0
≦3.0*
≦1.0
≦0.5
≦3.0
≦1.0
≦0.4
≦1.0
≦0.3
≦0.3

Cat. No
00-NPP-200

00-NPP-1000

00-NPP-5000

00-NPP-10000

Volume(µL)
20
100
200
100
500
1000
1000
2500
5000
1000
5000
10000

Accuracy
（%）
±1.0
±0.8
±0.8
±1.0
±0.8
±0.7
±1.0
±0.8
±0.6
±2.0
±0.8
±0.4

Precision
（%）
≦0.5
≦0.3
≦0.2
≦0.5
≦0.3
≦0.2
≦0.3
≦0.3
≦0.2
≦0.4
≦0.3
≦0.2

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NPP-2 below 0.2μL and NPP-10 below 1μL depend on the sampling skill and the sampling condition where pipettes are used.

Nichiryo's quality starter packages

S-PAC

ceramic plunger
Non corrosive ceramic used in
the plunger.( 100µL – 10000µL )■ Package type of 3 selected Pipettes with racked -Tips.

Materials used for pipette body
have high tolerance tosolvents and drop impact.

Tested600,000
times■ Ready-to-start packages with 3 sizes of

pipettes to perform highAC and CV pipetting.
■ 3 packages are available to cover wide range of volume.

Seal ring
Seal ring for the Premium istested dispensing / aspirating
600,000 times.
Non-greased plunger makes it
easy to maintain. ( 2µL – 200µL )Nozzle top has highly abrasion
tolerant. ( 2µL – 1000µL )The test was conducted by Nichiryo.
The test result may vary with conditionof usage.

Nichiryo Tube Opener (code:00-NTO)

S-PAC
Racked Tip
3 pcs【Parts List】

Tube Opener

3 or 4 racks“Easier” and “safer” to open tube lid with holding pipette

3 pcs■ Attached on grip end, 0.5, 1.5 mL tube

and screw cap can be opened easily.
＊Compatible with Nichipet Premium ( 2-1000µL )and Nichipet EXⅡ ( 2-1000µL )● What's new!
Shortened to releasetension to your thumb.

■ Hand temperature doesnʼt permeate through the body of the pipettewhich works to avoid affecting volume measurement. ( Patented )＊All features of Nichipet EXⅡ are available on Nichipet Premium.
Since the volume indicator is made of glass, ultraviolet germicidal of the clean bench will not make it cloud.
From prolonged irradiation, the surface layer of plastic may deteriorate or discolor, but its performance is not affected.

09

page.
www.nichiryo.co.jp/en/

Shortened
the distance tooperate smoothly.

Nichipet Premium S-PACの構成
Description

Cat. No

Nichipet Premium
A
00-SP-AP
Nichipet
EX S-PAC
S-PACの構成

Content
NPP-10、NPP-100、NPP-1000
BMT2-SSWR（0.5〜10μL）、
BMT2-SGR（2〜200μL）、
BMT2-LGR（100〜1000μL）、
NTO

Nichipet Premium S-PAC B

00-SP-BP

NPP-20、
NPP-200、NPP-1000
BMT2-SGR（2〜200μL）
×2、
BMT2-LGR（100〜1000μL）、
NTO

Nichipet Premium S-PAC H

00-SP-HP

NPP-2、NPP-10、NPP-100
BMT2-UTWR（0.1〜10μL）
×2、
BMT2-SSWR（0.5〜10μL）、
BMT2-SGR（2〜200μL）、
NTO＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

10

Nichipet EX PlusⅡP I P E T TE

Specifications

Solvent Resistant Nichipet EXⅡ

Cat. No
00-NPLO2-2
00-NPLO2-1000-NPLO2-20
00-NPLO2-100
00-NPLO2-200
00-NPLO2-1000
00-NPLO2-5000
00-NPLO2-10000

Chemical Resistant Model of Nichipet Series

Volume setting

Volume range(µL)
0.1 – 2
0.5 – 102 – 20
10 – 100
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Digital setting

Increments (µL)
0.002
0.01
0.02
0.1
0.2
1
10
10

Usable tips
BMT2-UT/UTWR/UTWRB、FUT/FUTRB
BMT2-SSW/SSWR/SSWRB、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB
BMT2-X/XR/XRB
BMT2-Z/ZR/ZRB*See p.18 for tip selection.

Accuracy and Precision
Cat. No
00-NPLO2-2

00-NPLO2-1000-NPLO2-20

00-NPLO2-100
Materials used for pipette body
have high tolerance to solvents.
Non-grease designed. No need
to worr y about continuous
autoclaving.

Easy calibration with enclosed
jig.

Volume(µL)
0.2
1
2
1
5
10
2
10
20
10
50
100

Accuracy(%)±12.0*
±5.0
±3.0
±4.0*
±1.0
±1.0
±5.0
±1.0
±1.0
±2.0
±1.0
±0.8

Precision(%)
≦6.0*
≦2.5
≦1.0
≦3.0*
≦1.0
≦0.5
≦3.0
≦1.0
≦0.4
≦1.0
≦0.3
≦0.3

Cat. No
00-NPLO2-200

00-NPLO2-1000

00-NPLO2-5000

00-NPLO2-10000

The chemical resistance of Perfluoro rubber O-ring▷A perfluoro rubber O-ring is installed in Nichipet EX plusⅡ.

Chemicals

Chemical-resistant O-ring withminimum expansion ratio.

▷PTFE ( Fluorine resin ) in the air tight chamber of the instrument.
▷A Perfluoro rubber O-ring with low cubical expansion in the air tight parts.
▷All models have plungers made in materials resistant to organic solvents.
■ Hyper Blower System; a longer push stroke from the second stop position, to improve dispensing efficiency. (2μL, 10μL)■ Easy-Calibration function provides simple lab calibration.
■The one-touch locking mechanism makes it easy to lock the sample volume.

＊All features of Nichipet EXⅡ are available on Nichipet EX PlusⅡ.
Since the volume indicator is made of glass, ultraviolet germicidal of the clean bench will not make it cloud.
From prolonged irradiation, the surface layer of plastic may deteriorate or discolor, but its performance is not affected.

11

page.
www.nichiryo.co.jp/en/

Precision(%)
≦0.5
≦0.3
≦0.2
≦0.5
≦0.3
≦0.2
≦0.3
≦0.3
≦0.2
≦0.4
≦0.3
≦0.2

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NPLO2-2 below 0.2μL and NPLO2-10 below 1μL depend on the sampling skill and the sampling condition where pipettes are used.

▷The volume growth of Perfluoro rubber against acid, alkalis,ammonia water, ester, furan and amine is small.

■ Advanced durability against organic solvent dispensing.

Accuracy(%)
±1.0
±0.8
±0.8
±1.0
±0.8
±0.7
±1.0
±0.8
±0.6
±2.0
±0.8
±0.4

*The Nichipet EX PlusⅡ is made of parts resistant to organic solvents which makes it more durable than previous pipettes in environments with organic solvents. However, it does notguarantee accuracy of dispensing organic solvents.
*As the body of the Nichipet EX PlusⅡ is not resistant to organic solvents, make sure it does not come in contact with the organic solvents.

C eramic plunger is used toprevent from corrosion bysolvents. ( 200uL – 10000uL )【Parts List】

Volume(µL)
20
100
200
100
500
1000
1000
2500
5000
1000
5000
10000

Resin material of ejector pipe
could enhance high tolerance tosolvents.

Temperature

Inorganic acid & Organic acid
Glacial acetic acid40℃
Acetic anhydride
40℃
Inorganic alkalies
28% Ammonia water
40℃
Furan & Aldehyde
Tetrahydrofuran
40℃
2-Methyl Tetrahydrofuran
40℃
Acetaldehyde
25℃
Acetophenone
40℃
Inclusive nitrogen compound
Acrylic nitrile
40℃
Ethylene diamine
40℃
Pyridine
40℃
N, N - dimethylformamide
40℃
N, N - dimethylacetamide
40℃
N-methyl-2-pyrrolidone
100℃
Carbon hydride, Carbon hydride halide
Toluene40℃
Xylene40℃
Chloroform
40℃
Alcohols
Methyl carbitol
40℃
Ethyl carbitol
40℃
Others, Oil, Steam
DN cut HS-1 cutting oil
130℃
O-148LCT Air craft engine oil
175℃

Test days

Perfluoro

Fluorine rubber
normal O-ring

21
21

A
A

C
C

21

A

D

21
21
21
21

C
B
B
A

D
D
D
C

21
21
21
21
21
7

A
B
A
A
A
A

C
D
C
D
D
D

21
21
21

Ａ
Ａ
Ｂ

Ｂ
Ｂ
Ｂ

21
21

Ａ
Ａ

Ｂ
Ｂ

11
20

Ａ
Ａ

A
Ｂ

A : Volume expansion rate below 5% …………
B : Volume expansion rate 5% - 20% …………
C : Volume expansion rate 20% - 50% …………
D : Volume expansion rate over 50% …………

No problem at all
No problem
Possible with some conditions
Nonusable

Chemicals

Temperature

Test days

Perfluoro

Fluorine rubber
normal O-ring

Ketone, Ester, Ether
Acetone
Methyl ethyl ketone
Methyl isobutyl ketone
Isophorone
Diacetone alcohol
gamma - Butyrolactone

40℃
40℃
40℃
40℃
40℃
40℃

21
21
21
21
21
21

B
A
Ｂ
Ａ
Ａ
Ａ

D
D
Ｄ
Ｄ
Ｃ
Ｃ

Diethylene carbonate
Acetylacetone
Methyl formate
Methyl acetate
Acetic ether
Isoamyl acetate
Methyl acetoacetate
Ethyl acetoacetate
Acrylic acids
Ethyl acrylate
Diethyl oxalate
Triethyl phosphate
Tricresyl phosphate
Diethyl ether
1, 4 - dioxan
Methyl - t - butyl ether

40℃
40℃
25℃
40℃
40℃
40℃
40℃
40℃
40℃
40℃
40℃
40℃
100℃
25℃
40℃
100℃

21
21
21
21
21
21
21
21
21
21
21
21
7
21
21
7

A
A
B
A
B
A
Ａ
Ａ
A
A
A
A
A
B
Ａ
Ａ

D
D
D
D
D
D
Ｃ
Ｃ
C
D
D
D
C
C
Ｃ
Ｃ

＊For more information, please feel free to contact us.

＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

12Nichipet EXⅡP I P E T TE

Specifications

Autoclavable Micro Pipette

Cat. No
00-NPX2-2
00-NPX2-1000-NPX2-20
00-NPX2-100
00-NPX2-200
00-NPX2-1000
00-NPX2-5000
00-NPX2-10000

Best Selling Model of Nichipet Series

Volume setting

Volume range (µL)
0.1 – 2
0.5 – 102 – 20
10 – 100
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Digital setting

Increments(µL)
0.002
0.01
0.02
0.1
0.2
1
10
10

Usable tips
BMT2-UT/UTWR/UTWRB、FUT/FUTRB
BMT2-SSW/SSWR/SSWRB、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB
BMT2-X/XR/XRB
BMT2-Z/ZR/ZRB*See p.18 for tip selection.

Accuracy and Precision
Cat. No

Volume(µL)
0.2
1
2
1
5
10
2
10
20
10
50
100

00-NPX2-2

00-NPX2-1000-NPX2-20

00-NPX2-100
Volume display made of durable
gla s s materia l prote cts thedamage from UV irradiation andkeep the pipette safer in clean
bench even over a long time.

Designed with a lock mechanism,
it is easier to adjust volume
smoothly.

Accuracy(%)±12.0*
±5.0
±3.0
±4.0*
±1.0
±1.0
±5.0
±1.0
±1.0
±2.0
±1.0
±0.8

Precision(%)
≦6.0*
≦2.5
≦1.0
≦3.0*
≦1.0
≦0.5
≦3.0
≦1.0
≦0.4
≦1.0
≦0.3
≦0.3

Cat. No
00-NPX2-200

00-NPX2-1000

00-NPX2-5000

00-NPX2-10000

Volume(µL)
20
100
200
100
500
1000
1000
2500
5000
1000
5000
10000

Accuracy(%)
±1.0
±0.8
±0.8
±1.0
±0.8
±0.7
±1.0
±0.8
±0.6
±2.0
±0.8
±0.4

Precision(%)
≦0.5
≦0.3
≦0.2
≦0.5
≦0.3
≦0.2
≦0.3
≦0.3
≦0.2
≦0.4
≦0.3
≦0.2

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NPX2-2 below 0.2μL and NPX2-10 below 1μL depend on the sampling skill and the sampling condition where pipettes are used.

Nichiryo's quality starter packages

The ejector pipe fixed by ascrew is easier to remove.
Materials used for pipette body
have high tolerance to solvents.
Non-grease designed. No need
to w o r r y a b o u t c o nt i n u o u s
autoclaving.

S-PAC■ Package type of 3 selected Pipettes with racked -Tips.
■ Each package contains 3 Nichipet EXⅡs, 3 sample racked

tips and 3 tube openers.

Resin material of ejector pipe
could enhance high tolerance tosolvents.

Ⅱ
S-PAC

Non corrosive ceramic used in
the plunger. ( 200uL – 10000uL )3 pcs【Parts List】■ Hyper blower system, longer second-push stroke, improves dispensing efficiency.（ 2μL,10μL ）Racked Tip

Tube Opener

3 or 4 racks

3 pcs

Nichipet Premium S-PACの構成■ Easy and convenient single hand locking lever.
■ PTFE ( Fluorine resin ) is used in the airtight chamber of the instrument, so the instrument keeps its air tightness

and precise reproducibility for long hours.
■ Filtered nozzles are used for volume models. ( 1000µL – 10000µL )■Tip ejector allows for tip ejections without touching the tip.
■Tip ejector is made of plastic to avoid damaging glass tubes.

Since the volume indicator is made of glass, ultraviolet germicidal of the clean bench will not make it cloud.
From prolonged irradiation, the surface layer of plastic may deteriorate or discolor, but its performance is not affected.

13

page.
www.nichiryo.co.jp/en/

Nichipet EXⅡ S-PACの構成
Description

Cat. No

Nichipet EXⅡ S-PAC A

00-SP-A2

NPX2-10、
NPX2-100、NPX2-1000
BMT2-SSWR（0.5〜10μL）、
BMT2-SGR（2〜200μL）、
BMT2-LGR（100〜1000μL）、
NTO

Content

Nichipet EXⅡ S-PAC B

00-SP-BN2

NPX2-20、
NPX2-200、
NPX2-1000
BMT2-SGR（2〜200μL）
×2、
BMT2-LGR（100〜1000μL）、
NTO

Nichipet EXⅡ S-PAC H

00-SP-H2

NPX2-2、
NPX2-10、
NPX2-100
BMT2-UTWR（0.1〜10μL）
×2、BMT2-SSWR（0.5〜10μL）、
BMT2-SGR（2〜200μL）、NTO＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

14

NICHIRYO Le / Nichipet ECO

PI PE TTE

Light & Ergonomic Pipette

Glass Tip Pipette

Nichipet ECO®

Light weight and easy to use
even for first-time users.

Eco-friendly glass tips suitablefor "Environmental Hormone" research.

■ The enlarged and stream lined grip fits well with your fingers and can be kept firmly■ The accuracy and precision of a micropipette with the stability and reliability of glass tips.

in hand and used with much less stress.

■ The glass tip enables bias free liquid handling for endocrine disruptor's analysis.

■ The user-friendly and enlarged push-button can be pressed easily and used for a long

■ Four different sizes covering a wide volume range. ( 20µL – 10000μL )time with less risk of RSI.

■ Glass tip is secured to the unit with a double O-ring design.

■ The enlarged push button makes it easy to set the volume and there is no need

■ Comes with a PTFE nozzle and ceramic piston.

to lock the dial for immediate operation.

【Parts List】

Glass Tip

Specifications
Cat. No
00-NLE-10
00-NLE-20
00-NLE-100
00-NLE-200
00-NLE-1000

Volume setting

Digital setting

Volume range(µL)
0.5 – 102 – 20
10 – 100
20 – 200
100 – 1000

Increments (µL)
0.02
0.02
0.2
0.2
2

Usable tips
BMT2-SSW/SSWR/SSWRB、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB【Specifications】*See p.18 for tip selection.

Accuracy and Precision
Cat. No
00-NLE-10

00-NLE-20

00-NLE-100

15

page.
www.nichiryo.co.jp/en/

Volume(µL)
1
5
10
2
10
20
10
50
100

Accuracy(%)
±6.0*
±2.0
±1.2
±5.0
±2.0
±1.2
±2.0
±1.2
±1.0

Precision(%)
≦5.0*
≦2.0
≦0.8
≦3.0
≦2.0
≦0.8
≦1.0
≦0.8
≦0.6

Cat. No
00-NLE-200

00-NLE-1000

Volume(µL)
20
100
200
100
500
1000

Accuracy(%)
±1.2
±1.0
±0.8
±1.2
±1.0
±0.7

Precision(%)
≦1.0
≦0.8
≦0.6
≦0.8
≦0.6
≦0.5

*Each accuracy and precision are the values which come from measuring with our compatible tip.
*The AC and CV of NLE-10 below 1μL depend on the sampling skill and the sampling
condition where pipettes are used.

Specifications
Cat. No
00-NPC-200
00-NPC-1000
00-NPC-5000
00-NPC-10ML● Glass tips
Volume range(µL)
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Increment(µL)
0.2
1.0
10.0
50.0

Accuracy(%)±6.0 – ±3.0
±5.0 – ±3.0
±5.0 – ±3.0
±6.0 – ±3.0

Precision(%)≦6.0 – ≦3.0
≦5.0 – ≦3.0
≦5.0 – ≦3.0
≦6.0 – ≦3.0

Usable tips
Glass Tip S
Glass Tip L
Glass Tip X
Glass Tip Z

Cat. No00-GLT-S
00-GLT-L
00-GLT-X
00-GLT-Z＊visit our website for parts list.

Volume range(µL)
20 – 200
100 – 1000
1000 – 5000
1000 – 10000

Q'ty(box)
10
10
10
10

page.
www.nichiryo.co.jp/en/

16

Nichipet FⅡ&VⅡ / MODEL 3100 / Premium Tip

P I P E T TE

Fixed Volume & Three Volume Micro Pipette

Premium Tip
Tip■The light and slender design along with the smooth movement of

ニチペットプレミアム S-PACの内容

the plunger reduces the operatorʼs fatigue.
■ Built-in tip ejector■The fine design and finishing of the top of the tips reduce the surface tension of liquid■ PTFE ( Fluorine resin ) is used in the airtight chamber of the instrument,

and ensure the tip to be residue free.

so the instrument keeps its air tight seal and precise reproducibility for long hours.

■The excellent sealing and discharge ability enable high accuracy of pipetting.
■ Chemical and heat resistant, for which the advanced material “Highly purified

Nichipet FⅡ

polypropylene” is used.

▷Suitable when only one volume is needed.
▷A wide range of standard volumes from 10µL – 1000µL.
▷Other volumes within the range available for special order.

■ Fully autoclavable. ( 121℃ for 20 minutes ) Racks can be autoclaved as well.( *Exceeding

the recommended standard for autoclaving may cause deformities )■The tips being able to be attached directly from the racks avoids dirt and dust fromgetting in, thus rising the efficiency of the operations.

Nichipet VⅡ■ DNase/RNase free（ excludes X/Z series ）▷Three different volume models▷Volume change by simply turning the knob.

FⅡ■ UV resistant

VⅡ

Specifications

Specifications
Type

FⅡ

VⅡ

DNase
RNase

Disposable Tips

【Parts List】

Cat. No
00-NPF2-1000-NPF2-20
00-NPF2-25
00-NPF2-30
00-NPF2-50
00-NPF2-100
00-NPF2-200
00-NPF2-500
00-NPF2-1000
00-NPV2-S
00-NPV2-M
00-NPV2-L

Volume setting

Fixed

Three variablesettings

Volume range(µL)
10
20
25
30
50
100
200
500
1000
10/20/50
50/100/200
200/500/1000

Accuracy(%)
±1.2
±0.8
±0.8
±0.8
±0.6
±0.6
±0.6
±0.5
±0.5
±1.2 – ±0.7
±0.7 – ±0.5
±0.7 – ±0.5

Precision(%)
≦0.4
≦0.3
≦0.3
≦0.3
≦0.3
≦0.3
≦0.2
≦0.2
≦0.2
≦0.6 – ≦0.3
≦0.4 – ≦0.3
≦0.3 – ≦0.2

Usable tips
BMT2-SSW/SSWR/SSWRB、FSS/FSSRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB

Type
UT

SSW

BMT2-SG/SGR/SGRB、FSE/FSERB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB
BMT2-SG/SGR/SGRB、FSG/FSGRB
BMT2-SG/SGR/SGRB、FAG/FAGRB
BMT2-LG/LGR/LGRB、FLG/FLGRB

SG

SE*See p.18 for tip selection.

K

Variable Volume Macro Pipette

LG

MODEL 3100

LE

■ Four different volume models from 1 mL to 10mL

X

■ Light weight and slender body
■ Easy to operate■ Liquid touches only disposable tip

Z■ PTFE piston ensures complete sealing of the syringe.

ZS＊The tips are available for pipette models with in the volume range of the tip mentioned above.

Cat. No.

Specification

00-BMT2-UT

Bulk tip

00-BMT2-UTWR

Racked tip

00-BMT2-UTWRB

Racked tip（sterilized）

00-BMT2-SSW

Bulk tip

00-BMT2-SSWR

Racked tip

00-BMT2-SSWRB

Racked tip（sterilized）

00-BMT2-SG

Bulk tip

00-BMT2-SGR

Racked tip

00-BMT2-SGRB

Racked tip（sterilized）

00-BMT2-SE

Bulk tip

00-BMT2-SER

Racked tip

00-BMT2-SERB

Racked tip（sterilized）

00-BMT2-K

Bulk tip

00-BMT2-KR

Racked tip

00-BMT2-KRB

Racked tip（sterilized）

00-BMT2-LG

Bulk tip

00-BMT2-LGR

Racked tip

00-BMT2-LGRB

Racked tip（sterilized）

00-BMT2-LE

Bulk tip

00-BMT2-LER

Racked tip

00-BMT2-LERB

Racked tip（sterilized）

00-BMT2-X

Bulk tip

00-BMT2-XR

Racked tip

00-BMT2-XRB

Racked tip（sterilized）

00-BMT2-Z

Bulk tip

00-BMT2-ZR

Racked tip

00-BMT2-ZRB

Racked tip（sterilized）

00-BMT2-ZS

Bulk tip

00-BMT2-ZSR

Racked tip

00-BMT2-ZSRB

Racked tip（sterilized）

Volume range(µL)

Color

Q'ty(box)

Usable models

1000

0.1〜10

NAR-2、
10 / NAR-10M8、
10M12
NPP-2 / NLT-2
Natural960
NPX・NPX2-2 / NPLO・NPLO2-2c（96pcs.×10 /s） NP7-8V、
12V / NPM-8V、
12V

0.5〜10NaturalNPP-10 / NLT-10
NPX・NPX2-10 / NPLO・NPLO2-10960c（96pcs.×10 /s） NLE-10 / NPF2-10

2〜200

NaturalNAR-20、
100、
200 / NAR-100M8、
100M12、200M8、
200M12
NPP-20、
100、
200 / NLT-20、100、200
NPX2-20、100、200 / NPLO2-20、100、200
NLE-20、
100、
200 / NPF2-20〜200c
NPV2-S、
M / NP7-8S、
8L、
12S、
12L
（96pcs.×10 /s） NPM-8S、
8L、
12S、
12L

2〜200

Natural

1000

1000
960

1000
960

c

（96pcs.×10 /s）

NPX-20、
100、200
NPLO-20、
100、
200

1000

30〜300Natural100〜1000Natural100〜1000NaturalNAR-300M8、
300M12
NP7-8K、
12K
960
c
12K
（96pcs.×10 /s） NPM-8K、

1000

NAR-1000
NPP-1000 / NLT-1000
NPX2-1000 / NPLO2-1000c（96pcs.×10 /s） NLE-1000 / NPF2-500、1000 / NPV2-L

960

1000
960

c

（96pcs.×10 /s）

NPX-1000
NPLO-1000

200

1000〜5000

Natural1000〜10000

Natural1000〜10000

NaturalNAR-5000
NPP-5000
NPX・NPX2-5000c（50pcs.×2 /s） NPLO・NPLO2-5000/11DG-5000

100

200

NPP-10000
NPX・NPX2-10000
80c（40pcs.×2 /s） NPLO・NPLO2-1000020080NAR-10000c（40pcs.×2 /s）【Specifications】■ TIP（measure：mm）

Specifications● Disposable tips

Cat. No00-31N-1000
00-31N-2000
00-31N-5000
00-31N-10000● Accuracy and Precision

17

Volume range(mL)
0.2 – 1.0
0.4 – 2.0
1.0 – 5.0
2.0 – 10.0

Increment(mL)
0.1
0.1
0.5
0.5

Cat. No
00-MPT-1
00-MPT-2
00-BMT2-X
00-BMT2-Z

Volume range(mL)
0.2 – 1.0
0.4 – 2.0
1.0 – 5.0
2.0 – 10.0

Color
Clear
Clear
Clear
Clear

Q'ty(bag)
1000
200
200
200

± 1.0% / ≦ 1.0%

page.
www.nichiryo.co.jp/en/＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

18

NichimateAqua series / Pipette Mate NEO

D
ISPE
NS
ER
DISP
EN
SER

P I PE TTE

DNase
RNase

Pyrogen

Pipette Controller■ Nichimate AQUA Tip
Type

Length
31.3㎜

10μL
10μL
（Long）

48.0㎜

200μL

53.0㎜

1000μL

81.9㎜

Cat. No.

Specification

Q'ty(box)

00-AQA-N10

Bulk tip

1000

00-AQA-N10-RF

Refill tip

960pcs00-AQA-N10L

Bulk tip

00-AQA-N200

Bulk tip

1000

00-AQA-N200-RF

Refill tip

960pcs00-AQA-N1000

Bulk tip

1000

00-AQA-N1000-RF

Refill tip

960pcs■ User Friendly pipette controller, 167g only

（96pcs.× 10stacks）■ Long-life Lithium Ion battery is installed. About 1700 times

dispensing are possible by full charge ( 4 hours ).

1000■ Dial-typed adjustable speed controller for your choice.
■ UV resistant for clean bench work.

（96pcs.× 5stacks × 2）

■ Self- standing type without holder.
■ Fully autoclavable spare parts. ( Nosecone, Sylicon Holder,and Filter )（96pcs.× 5stacks × 2）

■ Complies with CE and RoHS.

■ Low Retention Tip
Type

LengthCat. No.
00-LRT-10

31.1㎜

10μL

10μL
(Long)

45.8㎜

200μL

50.8㎜

200μL
（Long）

91.1㎜

300μL

55.9㎜

1250μL

90.2㎜

Specification

Q'ty(box)

Bulk tip

1000

00-LRT-10-RF

Refill tip

960pcs00-LRT-10RB

Racked tip（sterilized）

00-LRT-10L

Bulk tip

00-LRT-10L-RF

Refill tip

00-LRT-10LRB

Racked tip（sterilized）

00-LRT-200

Bulk tip

00-LRT-200-RF

Refill tip

00-LRT-200RB

Racked tip（sterilized）（96pcs.×10 /s）

00-LRT-200LRB

Racked tip（sterilized）（96pcs.×2 /s）

00-LRT-300

Bulk tip

1000

00-LRT-300-RF

Refill tip

960pcs00-LRT-300RB

Racked tip（sterilized）

00-LRT-1250

Bulk tip

00-LRT-1250-RF

Refill tip

00-LRT-1250RB

Racked tip（sterilized）（96pcs.× 5stacks × 2）960c

（96pcs.× 10 /s）

1000
960pcs（96pcs.× 5stacks × 2）960c

（96pcs.× 10 /s）

1000【Specifications】

960pcs（96pcs.× 5stacks × 2）

960
192

c

c

（96pcs.× 5stacks × 2）960c

（96pcs.×10 /s）

1000
768pcs（96pcs.× 4stacks × 2）960c

（96pcs.×10 /s）■ Low Retention Filtered Tip
Type

LengthCat. No.

10μL

31.1㎜

00-LRT-F10RB

10μL（Long）

45.8㎜

00-LRT-F10LRB

50.8㎜

00-LRT-F100RB
00-LRT-F200RB

200μL
200μL
（Long）

91.1㎜

00-LRT-F200LRB

300μL

55.9 ㎜00-LRT-F300RB

1250μL19

page.
www.nichiryo.co.jp/en/

90. 2㎜00-LRT-F1250RB

Q'ty(box)

96000-LRT-F20RB

20μL
100μL

Specificationc（96pcs.× 10 /s）

Specifications
Cat. No
00-PMNEO

Racked tip（sterilized）
192c（96pcs.×2 /s）Batteries
Charger
Filter
Pipette Types960

c（96pcs.× 10 /s）

Description
Pipette Mate NEO
Lithium Ion（3.6V, 600mAh）
100/240V 50/60Hz,0.2A
20mm×0.2µm（Hydrophobic）
Plastic or glass from 1 to 100mL＊Plastics and glass pipettes not sold by NichiryoCat. No
00-PM-N005
00-PM-N013
00-PM-N100
00-PM-N024
00-PM-N030
00-PM-N025
00-PM-N026
00-PM-N027

Description
Head fitting
Tube holder
Filter, 0.2µm
Travel charger 5V US
Travel charger base unit
Travel charger UK adaptor*
Travel charger AU adaptor*
Travel charger EU＊UK and AU adaptors need a travel charger base unit.

＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

20

Nichimate Stepper / MODEL 2100DG / DISPETD I SP E N SE R

Variable Repetitive Syringe Dispenser

Hand-held Dispenser

Nichimate Stepper

MODEL 2100DG■ Dispensing from most types of liquid containers is possible.
■ Easy to use with fixed volume adjustment■ 40 variations of dispensing are enabled by 8 syringes and 5 stages.

■ Light weight design for one-hand operation■ Suitable for dispensing viscose liquids due to the positive displacement system.

■The glass cylinder and plunger are protected by plastic housings.

■ Sterilized syringes are also available.
■ Syringe is made of PP and Piston is of PE. ( Piston for NS-V is made of stainless. )【Specifications】

Specifications
Cat. No00-21DG-1
00-21DG-2
00-21DG-5
00-21DG-10

Volume range(mL)
0.2 – 1.0
0.4 – 2.0
1.0 – 5.0
2.0 – 10.0

Increment(mL)
0.01
0.01
0.02
0.1

Volume readout
0/20 – 1/00
0/40 – 2/00
1/00 – 5/00
02/0 – 10/0

＊Liquid contact parts ; PTFE, Tygon®, Polypropylene, Pyrex®Glass, Ceramic and Platinum Iridium.

Accuracy and Precision±1.0% / ≦0.2%

Syringes for Stepper【Specifications】

Variable Volume Bottle-top Dispenser【Dimensional drawing】

DISPET■Can be attached directly to all standard reagent bottles.

Specifications
Cat. No
00-NSTP■Comes with an adapter ( ø22mm ) for direct attachment tosmaller reagent bottles. ( 100mL> ) ( not DP-5 and DP-10 )Product name
Nichimate Stepper■Light weight and compact design for easy handling.
■Liquid contact parts are chemically resistant material.

Accuracy and Precision■Extra tube ( 220mm ) is included and can be cut into a more convenient size.
dial Nosyringe00-NS-V
0.05mL
00-NS-S
0.5mL
00-NS-M
1.25mL
00-NS-N
2.5mL
00-NS-L
5mL
00-NS-X
12.5mL
00-NS-Y
25mL
00-NS-Z
50mL
Maximum number of dispenses

1
1µL
10µL
25µL
50µL
100µL
250µL
0.5mL
1mL
48

2
2µL
20µL
50µL
100µL
200µL
500µL
1mL
2mL
23

3
3µL
30µL
75µL
150µL
300µL
750µL
1.5mL
3mL
15

4
4µL
40µL
100µL
200µL
400µL
1000µL
2mL
4mL
11

5
5µL
50µL
125µL
250µL
500µL
1250µL
2.5mL
5mL
8

Accuracy(%)
±5.0
±1.7
±1.6
±1.2
±2.0
±1.1
±1.2
±1.5

Precision(%)
≦8.0
≦1.8
≦2.1
≦1.7
≦2.4
≦1.7
≦1.2
≦1.7

＊NS-V requires use of BMT-UT tips. NS-Y/Z requires use of NS-ADP adaptor. Syringes are disposable.

● Syringes
Cat. No
00-NS-V
00-NS-S
00-NS-M
00-NS-N
00-NS-L
00-NS-X
00-NS-Y
00-NS-Z

21

Description
0.05mL syringe (with 100 BMT-UT tips)
0.5mL syringe
1.25mL syringe
2.5mL syringe
5mL syringe
12.5mL syringe
25mL syringe (with one attachment)
50mL syringe(With one attachment)

page.
www.nichiryo.co.jp/en/

Q'ty(box)
100
100
100
100
100
100
50
25

Cat. No
00-NS-SB
00-NS-MB
00-NS-NB
00-NS-LB
00-NS-XB
00-NS-ADPY
00-NS-ADPZ

Description
0.5mL sterilized syringe
1.25mL sterilized syringe
2.5mL sterilized syringe
5mL sterilized syringe
12.5mL sterilized syringe
Attachment for 25mL（for 25mL Syringes / 00-NS-Y）
Attachment for 50mL（for 50mL Syringes / 00-NS-Z）

Q'ty(box)
100
100
100
100
100
1
1

■The wetted parts are chemically resistant.

Borosilicate glass,fluorocarbon polymers,Ceramic, Platinum-Iridium

Specifications
Cat. No00-DP-1
00-DP-2
00-DP-500-DP-10

Volume range(mL)
0.2 – 1
0.4 – 2
1–5
2 – 10

Increment(mL)
0.05
0.1
0.1
0.25

Accuracy(%)
±1.0
±1.0
±1.0
±1.0

Precision(%)
≦0.2
≦0.2
≦0.2
≦0.2

Amber bottle(option)
200mL
200mL
350mL
1L

Width(mm)×high(mm)59 × 255
59 × 255
66 × 290
98 × 350＊The threads for all bottle sizes are the same.
＊Please add "B" at the end of Cat.No when a bottle is needed.

AdapterCat. No
DP-1118
DP-1119
DP-1122

Size
18mm
19mm
22mm

Cat. No
DP-1124
DP-1125

Size
24mm
25mm

DP-112828mm

Cat. No
DP-1133
DP-1138
DP-1145＊visit our website for parts list.

Size
33mm
38mm
45mm

page.
www.nichiryo.co.jp/en/

22

D
AC
I SP
CEESSON SERRYDISPET EXⅡ
Ⅱ / DISPET Mini / ACCESSORYA C C E S S OR Y

Pipette Stand

Autoclavable Bottle-top Dispenser

DISPET EX■ Autoclavable（ 121℃ for 20 minutes ）■ 3 sized adapters for fitting with various types of bottles.
■ Chemical resistant materials are used for liquid contact parts.

Cat. No …………… 00-NAR-STD
Storing capacity … 8pcs
Compatible models … Nichipet Air＊
Nichipet Air -Multi ChannelNichipet Premium
Nichipet Premium LT
NICHIRYO Le

Cat. No …………… 00-MLT-STD2（ABS）
Storing capacity … 6pcs
Compatible models … Nichipet premium LT
Nichipet Premium
Nichipet EXⅡ
Nichipet EX PlusⅡ
Nichiryo Le
Nichipet EXⅡ MULTI

Cat. No …………… 00-NLE-STD（ABS）
Storing capacity … 1pc
Compatible models … Nichipet premium LT
Nichipet Premium
Nichipet EXⅡ
Nichipet EX PlusⅡ
Nichiryo Le

Cat. No …………… 00-BM-STD（Acrylic resin）
Storing capacity … 6pcs
Compatible models … Nichipet premium LT
Nichipet Premium
Nichipet EXⅡ
Nichipet EX PlusⅡ
Nichipet FⅡ&VⅡ■ Adjustable Inlet tube for various types of bottles.
■ Liquid returning mechanism for minimizing the waste of reagents.
■The wetted parts are chemically resistant.

Borosilicate glass,fluorocarbon polymers,Ceramic,Platinum-Iridium【Specifications】

Specifications
Cat. No00-DPX2-20
00-DPX2-50
00-DPX2-100
00-DPX2-250
00-DPX2-500
00-DPX2-1000

Volume range（mL）
0.2 – 2.0
0.5 – 5.0
1.0 – 10.0
2.5 – 25.0
5.0 – 50.0
10.0 – 100.0

Increment（mL）
0.05
0.1
0.2
0.5
1
2

Accuracy（%）
±0.5
±0.5
±0.5
±0.5
±0.5
±0.5

Precision
（%）
≦0.1
≦0.1
≦0.1
≦0.1
≦0.1
≦0.1

ThreadAttached adapters

GL-45

GL-28、
GL-32、
GL-38

Size
32 / 28mm
45 / 32mm

Cat. No00-DPX-003C
00-DPX-003D

Cat. No …………… 00-88-TY
Q'ty ………………… 50pcs
Compatible models … Nichipet Air -Multi ChannelNichipet EXⅡ MULTI
MODEL 8800
Nichipet Air -Multi Channel-

Ergonomically Designed
Repetitive Dispenser
Nichimate Stepper

AdapterCat. No00-DPX-003A
00-DPX-003B

Tray

Dispenser Stand

Cat. No …………… 00-NS-STD（Acrylic resin）

Size
45 / 38mm
45 / 40mm

Storing capacity

2pcs＊Numbers right side of ʻ/ʼ are caliber size of the bottles.

Bottle-top Dispenser

DISPET Mini

Tube Opener

Volume adjustment toolCat. No …………… 00-NTO-001
Q'ty ………………… 3pcs
Compatible models … Nichipet premium LT
Nichipet Premium
Nichipet EXⅡ
Nichipet EX PlusⅡ■Suitable for repetitively micro volume dispensing.
■Light and compact model for easy-to-carry-out.
■Chemical resistant materials are used for liquid contact parts.

Cat. No …………… 00-NAR-AT
Q'ty ………………… 1pc
Compatible models … Nichipet Air
Nichipet Air -Multi Channel-■The wetted parts are chemically resistant.

Borosilicate glass,fluorocarbon polymers,Platinum-Iridium＊Bottles are available as optional parts.

Filter【Specifications】

For NPP-1000
Cat. No 00-NP-0601000
Q'ty …… 10pcs

For NPLOⅡ-1000
Cat. No……… 00-LO2-0601000
Q'ty……… 10pcs

For NPP,NPXⅡ,NPLOⅡ-5000,10000
Cat. No … 00-NX2-0615000
Q'ty ……… 100pcs

For NAR-5000,10000
Cat. No … 00-NAR-F15000
Q'ty ……… 10pcs

Specifications
Cat. No00-DPM-250
00-DPM-1000
00-DPM-2000

23

page.
www.nichiryo.co.jp/en/

Volume range（µL）
100 / 250
500 / 1000
1000 / 2000

Accuracy（%）
±2.0
±1.0
±1.0

Precision
（%）
≦0.4
≦0.2
≦0.2

Thread
GL-28
GL-28
GL-28

For NPXⅡ-1000
Cat. No 00-NX2-0601000
Q'ty …… 10pcs＊visit our website for parts list.

page.
www.nichiryo.co.jp/en/

24

C USTO M M ADE & OE M
N OT ES【Examples】

Custom-Made products

Custom-made Pipettes■ Talk to us when you wish to have pipettes or molding

products that you cannot find in the market.
■ Full Turnkey Contract starting from planning, developing,

manufacturing and maintaining products will be possible.
■ Modification to our existing product is also welcomed.
■ Our custom-made molding products include disposable tip,

micro plate, cell culture vessels and other plastic product.

Custom-Made Automatic Dispenser

NSS-1000 Basic■ Cost effective custom made automatic dispensing systems■ Stand alone dispensers as well as dispensing components to be integrated into larger

system units will be possible.
■ Working areas ranging from A4-sized desk top type to larger work station type will bepossible.
■ Software to meet customers' requirements concerning C-language and Visual Basic

will be possible.

Custom-made system example
Appearance

25

Name

Main features

Dilution system formechanical oil

Function to prevent contamination
with air pressure feed-in.
1000mm(W)×500mm(D)×500mm

Samplepreparation
system bydisposable tip

Simple dispensing of samples from test
tubes to micro plates using disposabletips.
500mm(W)×800mm(D)×150mm

Two channel microplate
dilution system

Dilutes samples on micro plates. Twochannels with disposable and washable
tips.
700mm(W)×450mm(D)×700mm

Automatic picking
system forElectrophoresis
gel

Determines sample spots fractionatedby electrophoresis through image
analysis and performs safe picking
with the rotation and suction feature.
756mm(W)×700mm(D)×750mm

Multiple
dispensing anddilution system

Automatic dispenser with two syringe
pumps for dispensing processes andliquid line cleaning.
800mm(W)×450mm(D)×700mm

Five channel highspeed
dispensing system

System for dispensing five different
liquids using five syringe pumps andfive probes.
800mm(W)×500mm(D)×700mm

Dispenser for PCR
preparation

Reformatting from four 96-well microplates to one 384-well micro plate.
500mm(W)×400mm(D)×500mm

Powder sample
dispensing system

High accuracy dispenses of small
powder samples with the suction anddischarging function. Can also beconnected to electronic balances or
combined with liquid dispensing.
720mm(W)×700mm(D)×550mm

Five channelhigh throughput
dilution system

Automatic tip interval span feature andstacker for tips and micro plates areincluded.
1000mm(W)×610mm(D)×500mm

page.
www.nichiryo.co.jp/en/

Appearance

Name

Main features

page.
www.nichiryo.co.jp/en/26