Non-hazardous area

Installation Notes RIA46

A

L/-, L/+L/-, L/+Note wiring scheme on device!
ANALOG OUT

D12016 026-++0...20mA
4...20mA
CH1 0...10V
2...10V
0...5VOC 30V/200mA+0...20mA
4...20mA
CH2 0...10V
2...10V
0...5V

Temperature range
Ta

ANALOG INI

U>1Vx4x80/4...20mA

U<=1V

MUS 2-Lx3+-x1 x2x80...10V =0/1...5V +±10V±30V

MUS 4-Lx4x1 x2x7 x8=+x4x2x8+I-±100mV...±1V

RTD-4...20mA

I
0/4...20mA

TCx5 x6 x7 x82-L

ASSOCIATED INTRINSICALLY SAFEClass I, Zone 0 [Ex ia] IIC
Class I, Zone 2 Ex nA[ia] IIC

ASSOCIATED NONINCENDIVEClass I / Div. 2 / Groups ABCDx7 x8+4-L
3-L-

Typ J,K,T,N...

Designed

Pfanzelt
Refer to protection notice
ISO 16016

2009-05-12
Date (yyyy-mm-dd)

12 04 00 112

Unit

RIA46

2009-05-11

Edge of working parts Geometrical tolerancing
ISO 2768-mH-E
ISO 13715

Part No.

-Dwg.rev.

A
Scale

1:1Format

A4-40°C … +60°C

Revision no.

W10618

Revision date (yyyy-mm-dd)

2010-06-08

NameMPTitle

CONTROL DRAWING CSAAIS, ANI, NI

Material

71540268

DDVolume (mm³)

Drawing No.

Date (yyyy-mm-dd)-40°C ... +60°C

x4

T4

Pfanzelt

CC

D11 015 025

NONINCENDIVE Field WIRING INSTALLATIONClass I / Div. 2 / Groups ABCD
- The device is an Associated Nonincendive Safe equipment and must be installed in Division 2 or nonhazardous Locations only.
- The Nonincendive Field Wiring Circuit Concept allows interconnection of Nonincendive Field WiringApparatus with Associated Nonincendive Field Wiring Apparatus or Associated Intrinsically Safe
Apparatus or Associated Apparatus not specifically examined in combination as a system using any of thewiring methods permitted for unclassified locations, when Voc £ Vmax, Ca ³ Ci + Ccable, La ³ Li + Lcable.
- For entity installations use certified equipment that satisfy the following condition
Uo/Voc £ Vmax/Ui
Io/Isc £ Imax/Ii
Po £ Pi Co/Ca ³ Ci + Ccable Lo/ La ³ Li + LcableBB
CSA approved intrinsicallysafe apparatus

INTRINSICALLY SAFEClass I / Zone 0 [Ex ia] IIC- The device is an Associated Intrinsically Safe equipment and must be installed in Division 2 or nonhazardous Locations only.
- Installation should be in accordance with the Canadian Electrical Code (CEC).
- For entity installations use certified equipment that satisfy the following condition
Uo/Voc £ Vmax/Ui
Io/Isc £ Imax/Ii
Po £ Pi Co/Ca ³ Ci + Ccable Lo/ La ³ Li + Lcable- The Terminal of the intrinsically safe circuit must be placed at least a distance of 50mm from terminals of
the non intrinsically safe circuits, or adequate separators (e.g. ground metal partitions) must be used.

ACSA Approved Apparatus must be installed in accordance with manufacturer’s instructions.
Depending on location install per National Electrical Code (CEC) using wiring methods.
Use supply wires suitable for 5°C above surroundings.
For Class II keep tight when circuits alive.
The unit is installed in Class I, Division 2 area with two I.S. output channels (1 & 2), with cables clearanceof 2 mm minimum.
- Warning: Substitution of components may impair suitability for Class I, Division 2.
-CSA approved intrinsicallysafe apparatus

Approved

5Class I, Division 2, Groups ABCDClass II, Division 1, Groups EFG
Class IIIClass I, Zone 2 Group IIC

432

1
Hazardous (Classified) Locations
Class II, Division 1, Groups EFG
Class IIIClass I, Zone 0 Group IIC

XA02310R/09/EN/01.20

SeriesObjekt version

Sheet

Endress + Hause r
WetzerNesselwang / Germany

1 of 2 GmbH+Co. KG

54321

Umax £ 250 VAC
Umax £ 30 DC

Imax £ 3A
Imax £ 3A

CDI interface for device configuration
Impulse or Current output
Terminal O15, O16 or O25, O26

0/4…20 mA
Um £ 250 V

Output collector
Terminal D11, D12Imax £ 200 mA
Um £ 30 VDC

B

Cl. I, Gps ABCD
Cl. II, Gps EFG, Cl. III
Cl. I, Zone 0, IIC

Voc £ Vmax
Ca ³ Ci + CcableIsc £ Imax
La ³ Li + LcablePo £ PiASSOCIATED NONINCENDIVE FIELD WIRING
I,II,III/2/ABCDEFG
Voc £ Vmax
Ca ³ Ci + Ccable
La ³ Li + Lcable

2-wire transmitter power supply:
Terminal 11, 14, 12, 18 or
Terminal 21, 24, 22, 28

Voc £ 27.3 VIsc £ 96.5 mA
Po = 659 mW

Group A, B resp. IIC
Group C, D resp. IIB, IIA

Ca = 80 nF
Ca = 675 nFPfanzelt
Volume (mm³)

Designed

Pfanzelt
Refer to protection notice
ISO 16016

Drawing No.

Date (yyyy-mm-dd)

2009-05-12
Date (yyyy-mm-dd)

12 04 00 112

Unit

RIA46

2009-05-11

Edge of working parts Geometrical tolerancing
ISO 2768-mH-E
ISO 13715

Part No.

-Ca = 80 nF
Ca = 675 nF

4-wire transmitter power supply:
Terminal 14, 18 or
Terminal 24, 28

Voc £ 27.3 V
Isc £ 5 mA
Po = 34.2 mWGroup A, B resp. IIC
Group C, D resp. IIB, IIA

Ca = 80 nF
Ca = 675 nF

temperature input (RTD, TC):
Terminal 15, 16, 17, 18 and 12, 14 or
Terminal 25, 26, 27, 28 and 22, 24

Voc £ 27.3 V
Isc £ 22.1 mA
Po = 151 mW

Group A, B resp. IIC
Group C, D resp. IIB, IIA

Ca = 80 nF
Ca = 675 nF

Current input:
Terminal 14, 18 or
Terminal 24, 28

Voc £ 27.3 V
Isc £ 5 mA
Po = 34.2 mWGroup A, B resp. IIC
Group C, D resp. IIB, IIA

Ca = 80 nF
Ca = 675 nF

Voltage input:
Terminal 17, 18 and 13, 18 or
Terminal 27, 28 and 23, 28

Voc £ 27.3 V
Isc £ 5 mA
Po = 34.2 mWGroup A, B resp. IIC
Group C, D resp. IIB, IIA

Ca = 80 nF
Ca = 675 nF

Dwg.rev.

-Scale1:1
Format

A4

Revision no.

-

La = 4.625 mH
La = 19.125 mH

La = 1.525 H
La = 6.325 H

La = 81.725 mH
La = 327.425 mH

La = 1.525 H
La = 6.325 H

La = 1.525 H
La = 6.325 H

71540268

Revision date (yyyy-mm-dd)

Name

Material--

XA02310R/09/EN/01.20

Title

CONTROL DRAWING CSAAIS, ANI, NI

D

DApproved

La = 4.125 mH
La = 17.025 mH

Group A, B resp. IIC
Group C, D resp. IIB, IIACC

Entity parameters for channel 1&2 – Only one connected at a time:

Voc £ 27.3 VIsc £ 91.1 mA
Po = 622 mW

B

ASSOCIATED INTRINSICALLY SAFE

4-wire transmitter power supply:
Terminal 11, 12 or
Terminal 21, 22A

Output circuit limit relays
Terminal R12, R11, R13 orR22, R21, R23

5U £ 24…230 V AC/DC (-20%/+10%) 50/60 Hz

4321A

Power supply
Terminal L / +, L / -, PESeries
Objekt version

SheetEndress + Hause r
WetzerNesselwang / Germany

2 of 2 GmbH+Co. KG

54321