ONEBENCHTOP

AUTOCLAVABLEFERMENTER
BIOREACTORONE
The system consists of 2L fermenter/bioreactor (totalvolume), single wall glass vessel, bench-top, pre-assembled
unit, supplied with all necessary tubes, valves andinstruments, automation, control panel (software license).
The system is designed for aerobic and anaerobic
cultivations/ fermentations, closed asepticoperations.

No one liketheOne
Integrated wifi connection

Fully automatedAccurate stirring, temperature, pH and oxygen controls
Precise feedings via peristaltic pumps
Multiple use available up to 24 units managed in parallel

Applications
Process development
and optimization

Education

Basic Research Scale up and scale-down Small production
studiesRushton, Pitched Blade or Marine
impellersToro or Sintered sparger
Single-wall borosilicate glass
vessel, with thermoregulation
performed through heating blanket
and cooling finger.

Accurate and powerful rpm control,from 1 to 1900 rpm
Modbus digital sensors reduce
background noise and guarantee
quick response time.

Measurements and control options
included: stirring, temperature, pH,
dO₂
Suitable for batch, fed-batch andcontinuous processes

Compact stainless-steel PCS
equipped with 4 Watson Marlow
peristaltic pumps

Gas control through TMFC

Connectivity and data exchange via
in-built WiFi system
Multiple use available up to 24 units
managed in parallel

Leonardo

Innovative SCADA software​L​ EONARDO​: a smart and user-friendly
controller designed to provide a high level of automatedmanagement of the fermentation/cultivation processes. Multiple
use available up to 24 units managed in parallel

Workflowcustom phase manager

Synoptic
real time 3D view

Calibration
up to three-point

Trendscustom acquisition timeparallel visualizationcascade settings
peristaltic pumps function
assignable from software

calibration
simoultaneus calibration
values for parallel work

parallel control
manual control

up to 6 values
simultaneously display
automatic graphcomparison

devices in one

VesselSolaris Code
Production CodeTotal Volume (L)Ratio D/H
Min. Working Volume (L)Max. Working Volume (L)Max. temperature
Operating pressure
Headplate ports
DesignMaterials

Sensors length (mm)pH
dO₂

One 2.0
onest2.0
2.00
1:3.0
0,35
1,40
70 °C
< 0.5 barn.5 x M19 - n.4 x M16 - n.1 x M25
Borosilicate glass vessel (single wall)Borosilicate Glass and AISI 316 L

325
325

Dimensions for autoclave (with Condenser)
610
Height (mm)275
Diameter (mm)Stirring
DriveSpeed (rpm)Nominal Torque (Nm)
Impellers
Thermoregulation
Control
Total heater power (W)Brushless Motor
1-19000,9Select from: Rushtons impellers,Marine impellers, Pitched blade
PID Control - Accurancy 0,1 °C
400

Gas Control & Gas Mixing
Gas Control (Air)Sparger typeGas Out
Peristaltic Pumps
Typen.1 TMFC for Air
Select from: Toro type (ring), sintered microbubbling - bothprovided with 0,22 μm sintered filtern. 1 Condenser + 0,22 μm sinterized filter
up to n. 4 Watson Marlow type 114, fixed speed, max. 60 rpm,volumetric flow 0,5-51 ml/min, function assignable from software

Controller
Master Control Module
Leonardo software

35 x35 x 35 cm
LicenceTemperature
Sensor
Accuracy
Control system
Control range

PT100
0,1 °C
Measuring resident in Leonardo 3.2 software
0 - 70 °C

pH

SensorSensitivity
Control system
Control range
Operation temperature
Pressure range

dO₂Sensor
Accuracy
Control system
Control range
Operation temperature
Pressure range

Digital sensor
57 to 59 mV/pH
Measuring resident in Leonardo 3.2 software
0 - 14 °C
0 - 130 °C
0 - 12 bar
Digital Optical sensor±0.05%-vol, 21±0.2%-vol, 50±0.5%-vol
Measuring resident in Leonardo 3.2 software
0,05 - 300% air saturation-10 - 130 °C
0 - 12 bar