Operating ManualDiﬀerential Pressure Manometer
MODEL KS3200

Krone Corporation
HQ : post code 124-0023
9-6,Higashishinkoiwa 3 Chome, Katsushikaku Tokyo Japan
TEL :+81-3-3695-5431 / FAX : +81-3-3695-5698
E-Mail : sales-tokyo@krone.co.jp

KS3200
1. Out LineThe KS3200 is a very small sized digital diﬀerence pressure indicator which incorporates a monolithic silicon semiconductor pressure sensor.
It is the best solution to very low pressure measurement.
KS3200 is designed for the application of the semiconductor device equipment such as Coetar Developer,Cleaning and Furnace etc and has two transistor output(NPN or PNP) and Relay output, analog output 4-20mA or
1-5V.
Connector type XH connector (JST)

2. Handling Note
Recheck the following items before turning on the power.
Input and output wires, If the output terminal is connected to the power the interior circuit will be damaged.
Power voltage : Pressure range : Begin actual operation after the heat line is activated for over ﬁve minutes
after the power is turned on.
Excessive pressure.
Please do not apply pressure exceeding the maximum pressure as shown in the speciﬁcation on catalogues.
The excessive pressure may aﬀect the sensor characteristics and may make accurate measurement impossible.
Insert and contact of foreign matter.
A pressure sensor chip is placed inside the pressure port. If foreign matter such as wire inserts throughthe pressure port, damage could be occur.
This must be absolutely avoided.

• On this manualThis Manual serves to illustrate the capabilities and appropriate use of the Pressure Controller KS3200.
Inappropriate use or improper handling can lead to potential hazards for persons and intrinsic values.
Therefore, any person entrusted with working with the device must be instructed and aware of the potential risks.
This manual and its safety notices in particular must be addressed to with care.
If any doubts occur concerning the proper understanding of any paragraph of this manual, please consult themanufacturing company.
• If the equipment is used in a manner not speciﬁed by the manufacturer, the protection provided by the equipment
may be impaired.
• Do not close the pressure input ports when shipping, as changes in barometric pressure could damage instrumentswith low measuring ranges.
• Only technical personnel who are appropriately trained and authorized by the operator of the facility may assemble
the instrument and set up its electrical connections.
• The instrument may only be operated by appropriately trained individuals who have been authorized by the opera
tor of the facility.
• Measurement errors may occur if the instrument is not kept protected from sunlight.
• Altitude up to 2000 m;• Applicable POLLUTION DEGREE of the intended environment (POLLUTION DEGREE 2).
• Please do not apply power when connect the connector.
• Please do not apply leteral stress on the ports in order not to break the ports.
• Do not operate the switches on the back. If you move it by mistake, move it to the G side and then turn the powerback on.

-1-KS3200
3. Size
3-1 B8B-XH-A (8pins connector)
Dimensions : mm

1Port guard

Size for Panel cut

Panel thickness 0.5 to 4mm

KS3200Size for Front cover-2-8

KS3200
4. Speciﬁcation
ModelKS3200

Method of pressureDiﬀerentialSensor

Semi conductor piezoresistors wheatstone bridge

Pressure range

0-100〜5000Pa、±100〜±5000Pa

Display accuracy (*1)

±1%FS ±1digit,（More than 500Pa）
100Pa=±3%FS ±1digit
200Pa=±2%FS ±1digit
300Pa=±1.5%FS ±1digit

Temperature drift

Max ±0.1%F.S./℃（0〜50℃）

MediumAir, non-corrosive dry gasses (dust is not permitted)

Element

Semi conductor siliconOver pressure / Burst pressure

0-100〜1000、±100〜±1000：50kPa/80kPa
0-2000〜5000、±2000〜±5000：80kPa/550kPa

Power supply

24VDC(±10%）NEC(National Electrical Code) Class2 or LPS(Limited Power Source)

Current consumption

53 mA (within 20mA-Aout,Backlight:on, without PNP output current)

Setting method

Digital setting by front panel

Pressure setting range

Hi and Lo 0-100%F.S（min:1Pa)

Switching hysteresis

0-100%（min:1Pa)

Comparator output

Hi, Go, Lo output

Output[NPN open corrector]

Comparator

Sink Current:100mA less/ Resistive load(Vsat:1.35V)
Maximum applying voltage:24VDC[PNP transistor output]
Output voltage follows power source
Source current:80mA less (Voltage drops:2V)
Maximum applying voltage:24VDC[Relay output]*Contact: Normally open N.O (1-Form-A)
On-resistance: 0.25Ω (max)Dielectric strength voltage: 3750 Vrms (min)On-current: 0.9 A (max) resistiv load.

Analogoutput

Display

Maximum applied voltage: 24 VDC
Output display

7 segment LCD (with Backlight)

Display rate

Selectable from 0.1/0.5/1/2/5/10/20/30/60/180/300/600/1800/3600 sec.

Over ﬂow display

Display Blink at pressure in ≧110%F.S.

Analog output

4-20mA or 1-5V

Accuracy (*1)

±1.0%F.S〜

Load resistancestanderd=250Ω, Max : 350Ω

Resolution4μA*SSR(Solid State Relay) UL508-3-KS3200

environment

Item

Speciﬁcation

Operating temperature range

0-50 ℃（Not be freeze）

Humidity Limits

35-85%

Storage temperature range‑20℃〜+70℃

Operating environment

Indoor

Altitude

up to 2000 m

POLLUTION DEGREE

POLLUTION DEGREE 2

Protection standard

IP40 (Front panel only)

Warming up time

Not lower than 5 minuteSampling rate

0.1 sec

Insulation resistanceNot lower than 100ＭΩ（Across the panel and terminals、DC500Ｖ）

withstand voltage

500VAC（Across the panel and terminals、50/60Hz 1 minute)

Pressure port

max φ3.0mm (heat resistant polyamide)×2

Electric connection(8pins connector) JST：B8B-XH-A

Material

ABS(Case)

Dimensions

24(H)ｘ48(W)ｘ70(D)mm

Mounting

Panel mount

Mountng error

NoneWeight

Approx.48 g

Tests / Admissions

UL : E361798
CE : EN61326-1-2021
RoHS

Option Pressure Range and accuracy
Pressure Range

Display accuracy

Analog output accuracy

0-100 Pa

±3 %FS ±1digit±3 %FS

0-200 Pa

±2 %FS ±1digit±2 %FS

0-300 Pa±1.5 %FS ±1digit±1.5 %FS

0-500 Pa

±1 %FS ±1digit±1 %FS

0-1000 Pa±1 %FS ±1digit±1 %FS

0-2000 Pa±1 %FS ±1digit±1 %FS

0-2500 Pa±1 %FS ±1digit±1 %FS

0-5000 Pa±1 %FS ±1digit±1 %FS±100 Pa±2 %FS ±1digit±2 %FS±200 Pa±1.5 %FS ±1digit±1.5 %FS±300 Pa±1 %FS ±1digit±1 %FS±500 Pa±1 %FS ±1digit±1 %FS±1000 Pa

±1 %FS ±1digit±1 %FS±2000 Pa±1 %FS ±1digit±1 %FS±2500 Pa±1 %FS ±1digit±1 %FS±5000 Pa±1 %FS ±1digit±1 %FSFS : Full Scal

-4-

KS3200
5.Terminal description
B8B-XH-A (8pins connector)

1 2 3 4 5 6 7 8

JST：B8B-XH-A (8pins)

Port (＋)

18Port (−)

Pin number
Pin 1

Terminal name
Hi

Function contents
Hi Alarm output
NPN open corrector output, PNP transistor output
Relay output

Pin 2

Go (Good)

Go signal output
NPN open corrector output, PNP transistor output
Relay output

Pin 3

Lo

Lo Alarm output
NPN open corrector output, PNP transistor output
Relay output

Pin 4

GND (COM)

Power (common = GND)

Pin 5

A COM ( ‑ )Analog output (common)

Pin 6

A OUT ( + )Analog output (out)Pin 7

GND (COM)

Power (common = GND)

Pin 8V(+)

Power supply (DC24V)-5-KS3200
6. Application Schematic
B8B-XH-A (8pins connector)● NPN open corrector output+

24V

24V-

24V
123456784-20mA OutCHi

HiGoLo
COM
AOAO+
GND
24VDC

1RELAY2354

B8B-XH-A

B
CE

1B

Go

C

E

A

2V

1RELAY

354B

Lo

250

E

RELAY

2354

● PNP transistor output (option)

24V

124V
FUSE / 1A

2+E
E

E

Lo

1CB

Go

4-20mA OutRELAY

2354CBC

35424VBHiRELAY

123456781HiGoLoCOM
AOAO+
GND
24VDC

2RELAY

354A

B8B-XH-A
250

V*Analog output Recommendation Termination resistance
4-20mA : 250Ω (max. : 350Ω)
1-5V :10kΩ (min. : 1kΩ)*P1 : connector cable= Option-6-KS3200
6. Application Schematic
B8B-XH-A (8pins connector)

Control power supply

24V

24V

+

Power supply for KS3200+● Relay output ( + common)

-24V

-

24V

24V
12345678Vcc
4-20mA Out

1RL?

61HiGoLoCOM
AOAO+GND
24VDC

213HiB

1163CGo

1V

4RL?

RELAY

250EB

354354A

4RL?

RELAY

2B8B-XH-A

CRELAY23546E
34*Analog output Recommendation Termination resistance
4-20mA : 250Ω (max. : 350Ω)
1-5V :10kΩ (min. : 1kΩ)*P1 : connector cable= Option

CB

Lo

E

● Relay output ( ‑ common)
24V+

Control power supply

24V+

Power supply for KS3200

-24V

-

24V

24V
12345678Vcc
4-20mA Out

1RL?

6HiGoLoCOM
AOAO+GND
24VDC

12B8B-XH-A

3CHi

B

163CB

1V

4RL?

1RELAY

2354354250

EGo

A

4RL?

RELAY

12RELAY

3546E
34CLo

*Analog output Recommendation Termination resistance
4-20mA : 250Ω (max. : 350Ω)
1-5V :10kΩ (min. : 1kΩ)*P1 : connector cable= OptionBE-7-KS3200
7. Panel Mount
Installation<Front view><Side view>
Place the main body into the panel cut hole.
Mounting panel thickness is ....
Min. 0.5mm to Max. 4.0mm.

Mounting panel<Top view>
KS2900 to stick to panel through force of pressure.

-8-KS3200
8. Order code selection table
KS3200(ex. KS3200‒[1] ‒500 Pa

[2]I

‒

[3]N

‒

[4]N

‒

[5]C‒

[6]N)

[1] : Pressure Range
100 Pa, 200 Pa, 300 Pa, 500 Pa, 1000 Pa, 2000 Pa, 2500 Pa, 5000 Pa±100 Pa, ±200 Pa, ±300 Pa, ±500 Pa, ±1000 Pa, ±2000 Pa, ±2500 Pa, ±5000 Pa, other[2] : Analog output
I = 4-20mA output
V = 1-5V output[3] : Output connector
N = B8B-XH-A(JST)[4] : Alarm output (Hi, Lo , Go output )N = NPN transistor output
P = PNP transistor output
R = Relay output[5] : Connector Cable
N = non
C = With Connector Cable(XH-8pins and cable, L=1m )[6] : Option
Standard : N = nonOption : G = With Port Guard-9-KS3200
9. Accessories
B8B-XH-A (8pins connector)
JST XH connector (8pins) UL Recongnized : E60389
Wire : L = 1m, UL2651 AWG#28, (8pins)*Please do not use it other than an attention designation article.

- 10 -KS3200
10. Operaing components
10-1 Display[1] 7 segment LCD[2] Hi LCD[3] Go LCD[4] Lo LCD[5] MODE Key[8] SET Key[6] SHIFT Key[7] UP/Zero Key
10-2 Key functions (basic)

Back to measurement mode
Shift
Change the value
Return key(Over 5 sec.) →Zero adjustment

Eg.
1min.

1min.

Hi setDisplay
Hi Blink

MeasureDisplay

Lo setDisplay
Lo Blink
SET Key＋SET Key

SET Key

During operation, display returns to measuring mode after 1 min. if no Key operation done.
During Hi or Lo setting mode, measuring function is going on.

- 11 -KS3200
11. Setting menu
Menu Display

DescriptionIndicates ” Alarm setting mode”
To set alarm point for Hi
１＿Ｈindicates Hi setting，１＿Ｌindicates Low setting.
Alarm condition of １＿Ｈor１＿Ｌis to be selected in the Condition setting mode.
Figure shifted byand set by

key.

Value can be set within range of −９９９９〜＋９９９９digits.
During setting operation, Hi LCD display lighting.
To set alarm point for Lo
２＿Ｈindicates Hi setting,２＿Ｌindicates Low setting.
Alarm condition of ２＿Ｈor２＿Ｌis to be selected in the Condition setting mode.
Figure shifted byand set by

key.

Value can be set within range of −９９９９〜＋９９９９digits.
During setting operation, Lo LCD display lighting.
Indicates ” Alarm condition setting mode”
To set alarm condition for Hi.
Set bykey.

１＿Ｈ：to set Hi alarm point.
１＿Ｌ：to set Low alarm point.
During setting operation, Hi LCD display lighting.
To set hysteresis for Hi output to return from On to Oﬀ.
Figure shifted by

key and set bykey.

During setting operation Hi LCD display lighting.
Mode change for Positive Logic or Negative Logic.
In the case of relay output, the output is normally open when it is 1̲1, and the output is normallyclosed when it is 1̲0.
To set alarm condition for Lo.
Set bykey.

２＿Ｈ：to set Hi alarm point.
２＿Ｌ：to set Low alarm point.
To set hysteresis for Lo output to return from On to Oﬀ.
Figure shifted by

key and set bykey.

During setting operation Lo LCD display lighting.
Mode change for Positive Logic or Negative Logic.
In the case of relay output, the output is normally open when it is 1̲1, and the output is normallyclosed when it is 1̲0.
Mode change for Positive Logic or Negative Logic.
In the case of relay output, the output is normally open when it is 1̲1, and the output is normallyclosed when it is 1̲0.
To set delay time for Hi and Lo output to move from Oﬀ to On or On to Oﬀ.
ｏｎ.ｄ ：To set delay time to move from Oﬀ to On.
ｏＦＦ.ｄ：To set delay time to move from On to Oﬀ.
Set bykey.

Delay time can be set within range of ０〜９９９９ sec.
Figure shifted by

key and set by

key.

Alarm active or non active mode
Backlight blink or not when alarm value on the display- 12 -KS3200
12. Command
Indicates ” Operation Condition setting mode”
Manual ZERO setting

ON / OFF mode（key)ZERO display under 5% FS (Adjustable)
Sampling rate 0.1sec.
Display rate selectable from
0.1/0.2/0.5/1/2/5/10/20/30/60/180/300/600/1800/3600 sec.（key)ZERO suppress ON / OFF
Smallest digit ZERO ON / OFF（key)Mode protection ON / OFF
Active or non active for Analog output.
Analog output mode
ＳＡＰＬ：Outputs a raw value every 0.1 seconds.
ｄｉＳＰ：Outputs the average value moved in synchronization with the refresh setting.
ＡＳＰＬ：Outputs the moving average value every 0.1 seconds.
Disconnection detection function
By turning on the function, you can detect whether a device is connected to the Aout orAcom terminal. If it is not connected, "i.Err" will be displayed on the screen.
+(and over 3sec.)

Sampling average mode Display showing result of setting average.

- 13 -

KS3200
13. Parameter table
Default㻹㼑㼟㼡㼞㼙㼑㼚㼠＋Selectable（ >2 sec ）㻰㼕㼟㼜㼘㼍㼥㻖㻕㻌㻿㼑㼠㻌㼢㼍㼘㼡㼑㻌㼙㼡㼟㼠㻌㻌㻴㼕㻌㼜㼛㼕㼚㼠㻌䠚㻌㻸㼛㻌㼜㼛㼕㼚㼠㻯㼛㼙㼜㼍㼞㼕㼟㼛㼚㼂㼍㼘㼡㼑㻌㻿㼑㼠㻴㼕㻌㼜㼛㼕㼚㼠㻌㼟㼑㼠 㻖㻕㻌㻸㼛㻌㼜㼛㼕㼚㼠㻌㼟㼑㼠 㻖㻕㻌㻖㻕㻌㼞㼡㼚㻌㼟㼍㼢㼑㼟㻌㼠㼔㼑㻌㼟㼑㼠㼠㼕㼚㼓㼟㻌㼍㼚㼐㻌㼍㼜㼜㼘㼕㼑㼟㻌㼠㼔㼑㼙㻌㼠㼛㻌㼠㼔㼑㻌㻯㼛㻹㻼㻌㼙㼑㼚㼡㻱㻿㻼㻌㼟㼍㼢㼑㼟㻌㼠㼔㼑㻌㼟㼑㼠㼠㼕㼚㼓㻌㼢㼍㼘㼡㼑㻌㼍㼚㼐㻌㼠㼞㼍㼚㼟㼕㼠㼕㼛㼚㼟㻌㼠㼛㻌㼙㼑㼍㼟㼡㼞㼑㼙㼑㼚㼠㻌㼙㼛㼐㼑㻚㻖㻕㻌㻯㼛㼙㼜㼍㼞㼕㼟㼛㼚㻯㼛㼚㼐㼕㼠㼕㼛㼚㻌㼟㼑㼠䠴㻌䠴㻌䠴㻌䠴䠴㻌䠴㻌䠴㻌䠴㻿㼑㼠㻌㼒㼛㼞㻌㻿㼔㼕㼒㼠㻛㼁㼜㻌㼗㼑㼥㻿㼑㼠㻌㼒㼛㼞㻌㻿㼔㼕㼒㼠㻛㼁㼜㻌㼗㼑㼥㻴㼕㻌㻰㼕㼞㼑㼏㼠㼕㼛㼚㻴㼕㻌㻴㼥㼟㼠㼑㼞㼑㼟㼕㼟㻴㼕㻌㻻㼡㼠㼜㼡㼠㻌㻸㼛㼓㼕㼏㻖㻕㻌㻝䠻㻝㻩㻌㼚㼛㼞㼙㼍㼘㻌㼛㼜㼑㼚㻝䠻㻜㻌㻩㻌㼚㼛㼞㼙㼍㼘㻌㼏㼘㼛㼟㼑䠴㻌䠴㻌䠴㻌䠴㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼠㻌㼒㼛㼞㻌㻿㼔㼕㼒㼠㻛㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻸㼛㻌㻰㼕㼞㼑㼏㼠㼕㼛㼚㻸㼛㻌㻴㼥㼟㼠㼑㼞㼑㼟㼕㼟㻸㼛㻌㻻㼡㼠㼜㼡㼠㻌㻸㼛㼓㼕㼏㻖㻕㻌㻞䠻㻝㻩㻌㼚㼛㼞㼙㼍㼘㻌㼛㼜㼑㼚㻞䠻㻜㻌㻩㻌㼚㼛㼞㼙㼍㼘㻌㼏㼘㼛㼟㼑䠴㻌䠴㻌䠴㻌䠴㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼠㻌㼒㼛㼞㻌㻿㼔㼕㼒㼠㻛㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻳㼛㻌㻻㼡㼠㼜㼡㼠㻌㻸㼛㼓㼕㼏㻰㼑㼘㼍㼥㻌㼀㼕㼙㼑㻻㼡㼠㼜㼡㼠㻌㻿㼑㼘㼑㼏㼠㻮㼘㼕㼚㼗㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻖㻕㻌㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻖㻕㻌㻳䠻㻝㻩㻌㼚㼛㼞㼙㼍㼘㻌㼛㼜㼑㼚㻳䠻㻜㻌㻩㻌㼚㼛㼞㼙㼍㼘㻌㼏㼘㼛㼟㼑㻖㻕㻌䠴㻌䠴㻌䠴㻌䠴 㻖㻕㻌㻻㼜㼑㼞㼍㼠㼕㼛㼚㻯㼛㼚㼐㼕㼠㼕㼛㼚㻌㻿㼑㼠㼆㼑㼞㼛㻌㼍㼐㼖㼡㼟㼠㻸㼛㼣㻌㻯㼡㼠㼠㼕㼚㼓㻌㻾㼍㼠㼑㻖㻕㻌㼛㻲㻲㻚㼐㻌㻦㻌㻾㼑㼍㼏㼠㼕㼢㼑㻌㼠㼕㼙㼑㻌㼍㼒㼠㼑㼞㻌㼘㼑㼟㼟㻌㼠㼔㼍㼚㻌㼟㼑㼠㻌㼢㼍㼘㼡㼑㻚㻌㻌㻌㻌㻌㻌㼛㼚㻚㼐㻌㻦㻌㻾㼑㼍㼏㼠㼕㼢㼑㻌㼠㼕㼙㼑㻌㼍㼒㼠㼑㼞㻌㼙㼛㼞㼑㻌㼠㼔㼍㼚㻌㼟㼑㼠㻌㼢㼍㼘㼡㼑㻚㻿㼍㼙㼜㼘㼕㼚㼓㻌㻾㼍㼠㼑䠄䠂䠅䠄㼟㼑㼏䠅䠴㻌䠊䠴㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼠㻌㼒㼛㼞㻌㻿㼔㼕㼒㼠㻛㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻰㼕㼟㼜㼘㼍㼥㻌㻾㼍㼠㼑䠄㼟㼑㼏䠅䠌䠊䠍㻌䡚㻌䠏䠒䠌䠌㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻖㻕㻌㻜㻚㻜㻌㼠㼛㻌㻡㻚㻜㻑㻌㼙㼍㼤㻚㼆㼑㼞㼛㻌㻿㼡㼜㼜㼞㼑㼟㼟㻸㼍㼟㼠㻌㻰㼕㼓㼕㼠㻌㻲㼕㼤㻼㼞㼛㼠㼑㼏㼠㼕㼛㼚㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻖㻕㻌㻭㼚㼍㼘㼛㼓㻻㼡㼠㼜㼡㼠㻌㻿㼑㼘㼑㼏㼠㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻹㼛㼢㼕㼚㼓㻭㼢㼑㼞㼍㼓㼑㻌㻯㼥㼏㼘㼑䠌㻌䡚㻌䠎㻌䠑㻌䠑 㻖㻕㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻖㻕㻌㼀㼔㼑㻌㼐㼕㼟㼏㼛㼚㼚㼑㼏㼠㼕㼛㼚㻌㼐㼑㼠㼑㼏㼠㼕㼛㼚㻌㼟㼑㼠㼠㼕㼚㼓㻌㼙㼑㼚㼡㻌㼕㼟㻌㼐㼕㼟㼜㼘㼍㼥㼑㼐㻌㼛㼚㼘㼥㻌㼒㼛㼞㻌㼏㼡㼞㼞㼑㼚㼠㻌㼛㼡㼠㼜㼡㼠㻌㼠㼥㼜㼑㻚㻯㼍㻚㼛㻲㻦㻌㻻㼜㼑㼞㼍㼠㼑㼟㻌㼣㼕㼠㼔㼛㼡㼠㻌㼐㼕㼟㼜㼘㼍㼥㼕㼚㼓㻌㻎㼘㻚㼑㼞㼞㻎㻌㼣㼔㼑㼚㻌㼘㼛㼍㼐㻌㼐㼕㼟㼏㼛㼚㼚㼑㼏㼠㼕㼛㼚㻌㼛㼏㼏㼡㼞㼟㻚㻯㼍㻚㼛㼚㻦㻌㻰㼕㼟㼜㼘㼍㼥㻌䇾㼘㻚㼑㼞㼞䇿㻌㼣㼔㼑㼚㻌㼘㼛㼍㼐㻌㼐㼕㼟㼏㼛㼚㼚㼑㼏㼠㼕㼛㼚㻌㼛㼏㼏㼡㼞㼟㻚㻖㻕㻌㻿㼑㼠㻌㼒㼛㼞㻌㻜㻜㻝䡚㻞㻡㻡㻌㼏㼥㼏㼘㼑㻚㻔㻜㻜㻝㻌㼙㼑㼍㼚㼟㻌㼣㼕㼠㼔㼛㼡㼠㻌㼙㼛㼢㼕㼚㼓㻌㼍㼢㼑㼞㼍㼓㼑㻚㻌㻕㻿㼑㼠㻌㼒㼛㼞㻌㻿㼔㼕㼒㼠㻛㼁㼜㻌㼗㼑㼥㻖㻕㻻㼡㼠㼜㼡㼠㻌㼠㼕㼙㼕㼚㼓㻖㻕㻌㻿㻭㻼㻸䠖㻻㼡㼠㼜㼡㼠㻌㼞㼍㼣㻌㼢㼍㼘㼡㼑㼟㻌㻌㻌㼍㼠㻌㻿㻼㻸㻚㼞㻌㼠㼕㼙㼕㼚㼓㻌㻌㻌㻌㻌㼐㼕㻿㻼䠖㻭㼢㼑㼞㼍㼓㼕㼚㼓㻌㼕㼟㻌㼍㼘㼟㼛㻌㼞㼑㼒㼘㼑㼏㼠㼑㼐㻌㼍㼚㼐㻌㼛㼡㼠㼜㼡㼠㻌㼍㼠㻌㼠㼔㼑㻌㼐㼕㼟㼜㼘㼍㼥㻌㼠㼕㼙㼕㼚㼓㻚㻌㻌㻌㻌㻭㻿㻼㻸䠖㻭㼢㼑㼞㼍㼓㼕㼚㼓㻌㼕㼟㻌㼍㼘㼟㼛㻌㼞㼑㼒㼘㼑㼏㼠㼑㼐㻌㼍㼚㼐㻌㼛㼡㼠㼜㼡㼠㻌㼍㼠㻌㻿㻼㻸㻌㼠㼕㼙㼕㼚㼓㻚㻿㼑㼘㼑㼏㼠㻌㼒㼛㼞㻌㼁㼜㻌㼗㼑㼥㻹㼑㼟㼡㼞㼙㼑㼚㼠- 14 -

KS3200
14. Warranty
KRONE warrants the Products to be free of defects in materials and workmanship for a period of one year from thedate of shipment.
If any models or samples were shown to Buyer, such models or samples were used merely to illustrate the general
type and quality of the Products and not to represent that the Products would necessarily conform to said models orsamples.
Any Products found to be defective must be shipped to KRONE with all shipping costs paid by Buyer or oﬀered toKRONE for inspection and examination.
Upon examination by KRONE, at its sole option, will refund the purchase price of, or repair or replace at no charge
any Products found to be defective.
This warranty does not apply to any defects resulting from any action of Buyer, including but not limited to improper
installation, improper interfacing, improper repair, unauthorized modiﬁcation, misapplication and mishandling, such as
exposure to excessive current, heat, coldness, moisture, vibration or outdoors air.
Components which wear are not warranted. Pressure port portion is thinner. Please be aware of the damage causedby overload.
KRONE is pleased to oﬀer suggestions on the use of its various Products.
They are only suggestions, and it is Buyer's responsibility to ascertain the ﬁtness of the Products for Buyerʼs intendeduse.
KRONE will not be responsible for any damages that may result from the use of the Products.

OM40154-0025-01 Dec.2023- 16 -