Semiconductor manufacturing line, HVAC control ...

KS3200NEW

DifferentialPressure Manometer

Ideal for various types of differential pressure measurement and control

Visibility ◎HighContrast

The NEW digital differential pressure gauge KS3200 retains the functionality and accu-LCD

racy of previous model the KS2900, and KS3200 has new functions with a current output

Text color: white

disconnection function and an isolated relay control function. The display is using ahigh-contrast LCD, and the front panel is based on black with an emphasis on design.
And it can be easy to install to your equipment from the front with a one touch.
You can select the most suitable pressure range for your application from a total of 16
pressure ranges: 0-100Pa … 5000Pa and ±100Pa … ±5000Pa. In addition, the analog* Actual size

output can be selected from two types: 4-20mA or 1-5V(Option). Various settings can bevery easily to using on the front buttons, and a variety of micro-differential pressure measurements and controls can be performed with KS3200.

NEW FANCTION

Product features
・High Contrast LCD・Energy saving design・Analog output 4-20mA, 1-5V (Option)
・Hysteresis and delay output control
・Hi/Go/Lo open collector output switches
・Fail safe desigh・Short warm-up time
・UL/CE/RoHS

Current output disconnect
detection

Isolated relay (isolation)

This function detects whether theThis function allows the power sup-・Filter clogging monitoring

KS3200 and the analog output con-plied to the isolated relay (isolation)

・Clean room pressure monitoring

nection device are connected. (Cur-body to be treated as a control signal

・Air conditioning system monitoring (HVAC)

rent output Only) If a disconnection isfor a separate system. As a result,・Exhaust gas pressure monitoring

detected, “i.Err” will be showed on

the equipment has a high resistancethe disply.

to failure and is characterized byfewer malfunctions even in noisy
environments.

Appilication(Semiconductor equipment)・Wind tunnel testingKS3200

Order code & specification[Order code]
Type① Pressure Range② Analog output③ Output connector④ Alarm output⑤ Connector Cable□□□□□□□

□

□

□

□

100Pa / 200Pa / 300Pa / 500Pa / 1000Pa /2000Pa / 2500Pa / 5000Pa

I ：4-20mA
V ：1-5VDC

N：B8B-XH-A

N：NPN
P ：PNPR ：Relay

N：None
C：Connector Cable

N ：None
G ：Port Guard[Differential]

KS3200（JST）[Compound]⑥ Option±100Pa / ±200Pa / ±300Pa / ±500Pa /±1000Pa / ±2000Pa / ±2500Pa / ±5000Pa

KS3200

１２３４５６500Pa

4-20mA

B8B-XH-A（JST）

NPN

With Connector Cable

None

KS3200-500Pa-I-N-N-C-N

ex. Order code

Overview of setting function

Hi areaPressure
Value

1. Analogue and NPN open corrector output

Measurement
Value

2. Hysteresis and Window Mode

On Point
Off Point

Hi point

3. Deley Output

Hysteresis area

4. Manual ZERO adjustment

ONC.dLyt

5. Display time ratetime

6. Simple Moving Average

OFFC.dLyt

7. Low Value Cut ModeON

Hi On/Off

8. ZERO Suppress
9. Other function available

ONOFFON

OFF C.dLy

OFFON

ON C.dLy

ONON

OFF

[Specification]
KS3200SIZE* Other pressure range on request

Pressure Range Differential

0-50000-25000-1000Accuracy ±1.0 % F.S. ±1.0 % F.S.

(Pa)

Compound

0-±5000

0-±2500

0-100( back )±1.0 % F.S. ±1.0 % F.S. ±1.5 % F.S. ±3.0 % F.S.
0-±1000

Accuracy ±1.0 % F.S. ±1.0 % F.S.

0-3000-5000-±5000-±3000-±100

KS3200±1.0 % F.S. ±1.0 % F.S. ±1.0 % F.S. ±2.0 % F.S.

Differential pressure mode by semidconductor element

Pressure Mode

Air and Non-corrosive dry gases onlyMedia
Display Accuracy

Above-mentioned accuracy ±1digit

Thermal specification

Less than ±0.1％F.S./℃
80 kPaPressure overload
Pressure port

80 kPa

50 kPa

50 kPaLess than

50 kPa±0.15％F.S./℃
50 kPa2XWVLGHGLDPHWHU0D[ĭPP[4-digit(Full) / LCD displayDisplay
PressureAlarm
Low-cut
Zero adjusting

Max. 5% F.S
Manual Zero *Carry out regularly.
4-20mA / 1-5VDC (Option)

Analog output
Set upPressure setting range
OutputLoad current (Max)
Applied power (Max)Power supply
Power consumption
Electric connection
Operating temperature

Digital set up
0 〜 100% F.S
NPN Transistor open corrector, Hi / Go / Lo 3points, PNP (Option), Relay
100mA (Vol = 1.35V)

Port Guard (option)● B8B-XH-A (JST) / XHP-8 (JST)24V ±10%53mA (In that case 20mA output and back-light on)
1. Hi Output
2. Go Output
3. Lo Output
4. com8-pin connector
0℃ ... +50℃ (Not be frieze)
35％ ... 85％ (Not be condensation)-20℃ ... +70℃

Warming up

Over 5 minutes

Installation

CONNECTOR

DC24V (±10%)

Storage temperature
IP codeΦ3.0

Hi / Go / Lo 3points

Alarm output

Operating humidity

4-digit LCD (Unit : Pa)
Hi / Go / Lo 3points LCD display

5. Analogue 6. Analogue +7. GND
8. 24V*Recommended tubeĳîXUHWKDQH8 5LJLGĳî3)$OPTION

IP40 (Indoors)
Panel mount (Cut dimension : 45.0(W) x 22.2(H))

WeightApprox. 48 gram

CE Marking

CE, UL, RoHS
*Stand for KS series+81-3-3695-5431
9-6,Higashishinkoiwa 3 Chome, Katsushikaku Tokyo Japan
https://www.krone.co.jp
sales-tokyo@krone.co.jp

Caution*Connection cable (1m)

To use this device accurately and safety,carefully read manual.
Specification may change without notes.
Please contact local sales office for products selection and pricing.

CG40154-0015-01

2024/4