Temperature calibrator TP Basic Marine
TP Basic // Dry block // RT...650 °C // RT...1202 °F

TP Basic MarineHighlights• Optimized for the requirements of the marine market• Extremely competitive• Smaller, lighter and technically better than all other comparable models on the market

TP Basic
The temperature calibrators of the TP Basic series are characterised by their efficiency and portability. The series consists ofdry block calibrators covering a wide temperature range and areused on site, e.g. in marine applications.

The easy operation, the integrated internal reference temperature sensor and the dry block calibration function ensure an extremely easy calibration process.

SIKA temperature calibrators
Temperature calibrators are used for the verification of the functionality and calibration of temperature measuring devices and temperature sensors. As the sole German manufacturer of these devices, we develop and produce our “Made in Germany” temperature
calibrators with a special focus on long-term reliability and utmost accuracy in combination with easy operation. We can rely on
more than 40 years of experience in doing this: SIKA’s first dry block temperature calibrator was launched all the way back in 1980.
Every SIKA temperature calibrator is meticulously tested for accuracy and stability. This is attested by our standard calibration
certificate, which we issue with every temperature calibrator, or by means of an optional DAkkS calibration certificate [Germanaccreditation body]. This is to guarantee that you receive a perfect product which can be traced back to national and internationaltemperature measurement standards.

Technical datasheet
04/2024 // V1.1

SIKA Dr. Siebert & Kühn GmbH & Co. KGStruthweg 7-9 // 34260 Kaufungen // Germany

1 // 4www.sika.net

Features
Easy operation• The TP Basic Marine can be operated with only three buttons:
Two arrow buttons for setting the target temperatureand one button for confirmation.
• Thus, temperatures can be set as easily as, for example, in
the air conditioning system in your car• Any operational errors can be nearly excluded. You do
not need any specifically trained staff or time-consuming
briefings

Small block with particularly small calibration inserts• The TP Basic Marine has a small block that provides extremely fast heat-up and cool-down times thanks to its low
thermal mass. This saves time and money with every calibration process• The particularly small calibration inserts are specially designed for temperature sensors in the marine sector. They
are available with two different borehole sizes as standard:Ø 3.5 mm and 6.5 mm. For Ø 13.5 mm, the calibrator issimply operated without a sleeve. These sizes cover mosttemperature sensors in the marine sector. Customised
borehole sizes are also available. The small size of the calibration inserts enables cost savings on accessories.

SIKA Gold Service
SIKA Gold Service provides a comprehensive service package for the regular recalibration of your temperature calibrator.
You will benefit from exclusive savings and discounts as well as special promotions reserved to SIKA Gold Service members.
• You will save 33% in the recalibration of your temperature calibrator• You will receive a 10% discount on any repairs that may become necessary• You will receive preferential invitations to product presentations, symposia,
practice days and exclusive training offers
Register now and benefit from the SIKA Gold Service: gold-service.sika.net

Technical datasheet
04/2024 // V1.1

GOLDSERVICE

SIKA Dr. Siebert & Kühn GmbH & Co. KG
Struthweg 7-9 // 34260 Kaufungen // Germany

2 // 4www.sika.net

Technical Data
TP Basic Marine
Temperature range

RT…650 °C

RT...1202 °F

Dimension of the calibration insert

Ø 13.5 x 110 mm (Calibration insert easy to replace)

Dry blockAccuracy±1.2 °C±2.16 °F

Temperature stability±0.2 °C

0.36 °F

Resolution of the temperature display

0.1 °C0.1 °F

Temperature units°C / °F

Reference temperature sensor

internal, fixed installation

Dimensions
 Width Height Depth

127 mm274 + 50 mm (Handle)
272 mmWeightApprox. 5.1 kg

Power supply

115 V or 230 V Nominal voltage (two variants)

Power consumption

Approx. 650 W

Display
Display and control

2.4 inch color display, operation via three capacitive keys

Approvals

Article numbers
Temperature Calibrator
TypePower supply

Article number

TP Basic Marine

230 V

EPMA650M131100

TP Basic Marine

115 V

EPMA650M131102

Calibration insert
Bore holes [mm]FunctionCalibration insert [mm]Material

Article number

Without bore hole

Dry block

Ø 13

Brass

EZ11013000MS01

1x Ø 3.5

Dry block

Ø 13

Brass

EZ11013035MS01

1x Ø 5.0

Dry block

Ø 13

Brass

EZ11013050MS01

1x Ø 6.5

Dry block

Ø 13

Brass

EZ11013065MS01

1x Ø 8.5

Dry block

Ø 13

Brass

EZ11013085MS01

1x Ø 9.5

Dry block

Ø 13

Brass

EZ11013095MS01

1x Ø 10.5

Dry block

Ø 13

Brass

EZ11013105MS01

1x Ø 11.5

Dry block

Ø 13

Brass

EZ11013115MS01

Calibration insert incl. 1 bore hole of choice

Dry block

Ø 13

Brass

Please indicate bore
hole in the order

Calibration certificate - Select your calibration certificates as needed
Each calibrator is already delivered with a standard calibration certificate (6 test points).

Article number

SIKA works calibration certificate (similar to standard calibration certificate + marking on the calibrator)

EKTPWP1FKT

DAkkS calibration certificate (3 test points + measurement uncertainty determination)

EKTPDAKKS1FKTEach additional test point DAkkS calibration certificate

EKTPDAKKSZUSP

SIKA Gold Service works calibration certificate

EKTPGOLDWP

SIKA Gold Service DAkkSEKTPGOLDDAKKS

Accessories

Article number

Transport bag

XE2422

Technical datasheet
04/2024 // V1.1

SIKA Dr. Siebert & Kühn GmbH & Co. KG
Struthweg 7-9 // 34260 Kaufungen // Germany

3 // 4www.sika.net

Overview of SIKA temperature calibrators
Our series: Basic. Solid. Premium.
• Dry block calibrators of the TP Basic series impress with
their uncomplicated operation and high cost-effectiveness.
They are particularly suitable for use on ships or in industrial applications.
• Equipped with a PC interface, the dry block calibratorsand calibration baths of the TP Solid series cover a wide
temperature range with high accuracy.
Temperature range• For the highest demands on accuracy and flexibility:
The dry-block and multi-function temperature calibratorsof the TP Premium series represent the pinnacle ofour technical development. Equipped with an integrated
touch screen, a PC interface, an external reference sensor
and integrated measuring instrument, this series offers
extreme accuracies for all calibration tasks.

Accuracy-55 °C ... 200 °C-67 °F ... 392 °F

Dry blockDry blockDry block±0.4 °C±0.2 °C±0.2 °C±0.72 °F
±0.36 °F
±0.36 °F

TP Basic
28 x 150
TP Solid
28 x 150
TP Premium 28 x 150

TP 17200
TP 17200S
TP 37200E.2-35 °C ... 155 °C-31 °F ... 311 °F

Calibration bath±0.1 °C±0.18 °F

TP Solid

60 x 170

TP M165S

Dry blockDry blockDry blockDry blockDry block ext.
Dry block int.
Air Shield Insert
Calibration bath
InfraredSurface
Dry blockDry block±1 °C±0.4 °C±0.2 °C±0.2 °C±0.2 °C±0.3 °C±0.07 °C
±0.1 °C±0.5 °C±1 °C±0.4 °C±0.2 °C±1.80 °F
±0.72 °F
±0.36 °F
±0.36 °F
±0.36 °F
±0.54 °F
±0.126 °F±0.18 °F
±0.9 °F±1.88 °F
±0.72 °F
±0.36 °F

TP Basic
TP Basic
TP Solid
TP Premium

28 x 150
28 x 150
28 x 150
28 x 150

TP 17165M
TP 17165
TP 17165S
TP 37165E.2

TP Premium 60 x 170

TP 3M165E.2

TP Basic
TP Solid

60 x 150
60 x 150

TP 17166
TP 17166S

Dry block±0.05 °C

±0.09 °F

TP Solid

7 x 6.5 x 150

TP 17ZeroDry block±1 °C±1.80 °F

TP Basic

18 x 150

TP 18200E

Calibration bath
Dry block ext.
Dry block int.
Air Shield Insert
Calibration bath, tub insert, ext.
Calibration bath, tub insert, int.
Calibration bath, direct filling, ext.
Calibration bath, direct filling, int.
InfraredSurface
Dry blockDry blockDry blockAir Shield Insert
InfraredSurface
Dry blockDry blockDry blockDry blockDry blockAir Shield Insert±0.2 °C±0.25 °C
±0.5 °C±0.08 °C
±0.35 °C
±0.53 °C
±0.18 °C
±0.46 °C
±0.5 °C±1°C±0.6 °C±0.3 °C±0.3 °C±0.2 °C±0.5 °C±1 °C±1 °C±0.8 °C±1.2 °C±0.4 °C±0.43 °C
±0.27 °C

±0.36 °F
±0.45 °F
±0.9 °F±0.144 °F±0.63 °F
±0.954 °F±0.324 °F±0.828 °F±0.9 °F±1.8 °F±1.08 °F
±0.54 °F
±0.54 °F
±0.36 °F
±0.9 °F±1.8 °F±1.8 °F±1.44 °F
±2.16 °F
±0.72 °F
±0.744 °F±0.486 °F

TP Solid

60 x 170

TP M255S

TP Premium 29 x 150

TP 37700E.2Dry block±1 °C±1.8 °F

TP BasicTP 18850E-35 °C ... 165 °C-31 °F ... 329 °F-30 °C ... 165 °C-22 °F ... 329 °F-10 °C ... 100 °C
14 °F ... 212 °F
RT ... 200 °C
RT ... 392 °F

RT ... 255 °C
RT ... 491 °F

RT ... 450 °C
RT ... 842 °F

RT ... 650 °C
RT ... 1202 °F
RT ... 700 °C
RT ... 1292 °F
RT ... 850 °C
RT ... 1562 °F

Features

Block dimensions

Function(RT=Room temperature)[Ø mm x depth mm]

Type

TP Premium 60 x 170

TP 3M255E.2

TP Basic
TP Solid

TP 17450
TP 17450S

60 x 150
60 x 150

TP Premium 60 x 150

TP 37450E.2TP BasicTP Basic
TP Basic
TP Solid

TP 17650M
TP 17650
TP Basic Marine
TP 17650S

28 x 150
28 x 150
13.5 x 110
28 x 150

18 x 100

Subject to technical modifications and errors

Technical datasheet
04/2024 // V1.1

SIKA Dr. Siebert & Kühn GmbH & Co. KG
Struthweg 7-9 // 34260 Kaufungen // Germany

4 // 4www.sika.net