LINEARIZER

SGL
SGLL
INSTRUCTION MANUALPreface

Thank you for purchasing our SGL and SGLL, Linearizer. This manual contains instructions
for the mounting, functions, operations and notes when operating the SGL and SGLL. To
prevent accidents arising from the misuse of this instrument, please ensure the operator
receives this manual.

Notes• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed, serious injury
or malfunction may occur.
• The contents of this instruction manual are subject to change without notice.
• Care has been taken to ensure that the contents of this instruction manual are correct, but if
there are any doubts, mistakes or questions, please inform our sales department.
• This instrument is designed to be installed on a DIN rail within a control panel. If it is not,measures must be taken to ensure that the operator does not touch power terminals or otherhigh voltage sections.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos Co., Ltd. is not liable for any damage or secondary damage(s) incurred as aresult of using this product, including any indirect damage.
SAFETY PRECAUTIONS (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on circumstances, procedures indicated by
Caution may result in serious
consequences, so be sure to follow the directions for usage.
which may lead to dangerous conditions and cause
Warning Procedures
death or serious injury, if not carried out properly.

CautionProcedures which may lead to dangerous conditions and cause
superficial to medium injury or physical damage or may degrade
or damage the product, if not carried out properly.

Warning• To prevent an electrical shock or fire, only Shinko or qualified service personnel may
handle the inner assembly.
• To prevent an electrical shock, fire, or damage to instrument, parts replacement may only
be undertaken by Shinko or qualified service personnel.

Safety Precautions• To ensure safe and correct use, thoroughly read and understand this manual before using this
instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment. Verify correct usage after purpose-of-use consultation with our agency or main
office. (Never use this instrument for medical purposes with which human lives are involved.)• External protection devices such as protective equipment against excessive temperature rise,etc. must be installed, as malfunction of this product could result in serious damage to thesystem or injury to personnel. Proper periodic maintenance is also required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring
due to the instrument being used under conditions not otherwise stated in this manual.

Caution with Respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the
manufacture of weapons of mass destruction (i.e. military applications, militaryequipment, etc.), please investigate the end users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
- 2 -Installation Precautions

CautionThis instrument is intended to be used under the following environmental conditions(IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of -10 to 55 (14 to 131 )that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85 %RH• No large capacity electromagnetic switches or cables through which large current is flowing• No water, oil or chemicals or the vapors of these substances can come into directcontact with the unit.
• When installing this unit within a control panel, please note that ambient temperature ofthis unit – not the ambient temperature of the control panel – must not exceed 55(131 ). Otherwise the life of electronic components (especially electrolytic capacitor)
may be shortened.
Note: Avoid setting this instrument directly on or near flammable material even though
the case of this instrument is made of flame-resistant resin.

Wiring Precautions

Caution• Do not leave bits of wire in the instrument, because they could cause a fire andmalfunction.
• When wiring, use a crimping pliers and a solderless terminal with an insulation sleevein which an M3 screw fits.
• Tighten the terminal screw using the specified torque. If excessive force is applied tothe screw when tightening, the screw or case may be damaged.
• This instrument does not have a built-in power switch, circuit breaker and fuse. It isnecessary to install a power switch, circuit breaker and fuse near the instrument.
(Recommended fuse: Time-lag fuse, rated voltage 250 V AC, rated current 2 A)
• For wiring of the AC power source, be sure to use terminals as described in this manual.
If the AC power source is connected to incorrect terminals, the unit will be burnt out.
• Do not apply a commercial power source to the sensor which is connected to the
input terminal nor allow the power source to come into contact with the sensor.
• When using DC voltage and current input, do not confuse polarity when wiring.
• Keep the input/output wires and power line separate.

Operation and Maintenance Precautions

Caution• Do not touch live terminals. This may cause an electrical shock or problems inoperation.
• Turn the power supply to the instrument OFF when retightening the terminal orcleaning. Working on or touching the terminal with the power switched ON may result
in severe injury or death due to electrical shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)• As the display section is vulnerable, be careful not to put pressure on, scratch or strike
it with a hard object.
- 3 -Characters used in this manual
Indication
Number, /Indication
Alphabet
Indication
Alphabet[: No character is indicated (unlit).]

-1

0

1

2

3

4

5

6

7

8

9

°C

°F

A

B

C

D

E

F

G

H

I

J

K

L

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

Contents
page
1. Model ................................................................................................................................. 5
1.1
Model name ............................................................................................................ 5
1.2
How to Read the Model Label ............................................................................... 6
2. Name and Functions ........................................................................................................ 7
2.1
Front Panel ............................................................................................................. 7
2.2
Display Section ...................................................................................................... 8
3. Mounting ........................................................................................................................... 9
3.1
External Dimensions (Scale: mm) ........................................................................ 9
3.2
Mounting to, and Removal from the DIN Rail .................................................... 10
4. Wiring ............................................................................................................................... 11
4.1
Lead Wire Solderless Terminal ............................................................................ 11
4.2
Circuit Configuration ............................................................................................ 11
4.3
Terminal Arrangement ......................................................................................... 12
4.4
Wiring .................................................................................................................... 13
5. Display Mode .................................................................................................................. 14
6. Setting Mode ................................................................................................................... 16
6.1
Display Transition in Setting Mode .................................................................... 16
6.2
Input Setting Mode............................................................................................... 18
6.3
Output Setting Mode ............................................................................................ 20
6.4
Linearization Setting Mode ................................................................................. 22
6.5
Instrument Setting Mode ..................................................................................... 31
6.6
Communication Setting Mode ............................................................................ 33
6.7
Custom Display Setting Mode ............................................................................ 35
6.8
Manual Mode ........................................................................................................ 35
7. Adjustment ..................................................................................................................... 36
7.1
Basic Operation of Adjustment........................................................................... 36
7.2
Adjustment ........................................................................................................... 36
7.2.1
Output Adjustment ....................................................................................... 36
8. Operation ........................................................................................................................ 37
8.1
Indication after Power-on .................................................................................... 37
8.2
Operation .............................................................................................................. 37
8.2.1
Input Indication Range ................................................................................. 37
8.2.2
Output Indication Range .............................................................................. 37
8.2.3
Input Disconnection Status ......................................................................... 38
8.2.4
Indication Time Setting ................................................................................ 38
8.2.5
Linearization Function.................................................................................. 39
9. Specifications ................................................................................................................. 40
10. Troubleshooting ........................................................................................................... 43
10.1 Indication .............................................................................................................. 43
10.2 Key Operation ...................................................................................................... 43
10.3 Operation .............................................................................................................. 43
11. Character Table............................................................................................................. 44- 4 -1. Model
1.1 Model name
SGL
SGL-□□□-□-□

Option0: No option needed
1: Multi-rotation trimmer
2: Moisture-proof treatment
3: Multi-rotation trimmer + Moisture-proof treatment
Power supply
0: 100 to 240 V AC
1: 24 V AC/DC
Output *2
Input *1
Series name
SGLL
SGLL-□□□-0-□

Option0: No option needed
1: Multi-rotation trimmer
2: Moisture-proof treatment
3: Multi-rotation trimmer + Moisture-proof treatment
Power supply
100 to 240 V AC
output *2
Input *1
Series name (with Serial communication)*1: Input
Code
A0(Built-in 50

A1

(250

A2

(50

A3
A4
A5
A6
A7
A8
A9

Input Type(250
Directcurrent

(62.5
(250
(100
(100
(1000
(10

Code

4 to 20 mA
shunt resistor)
4 to 20 mA
shunt resistor)
4 to 20 mA
shunt resistor)
0 to 20 mA
shunt resistor)
0 to 16 mAshunt resistor)
2 to 10 mA
shunt resistor)
0 to 10 mA
shunt resistor)
1 to 5 mAshunt resistor)
0 to 1 mAshunt resistor)
10 to 50 mA
shunt resistor)- 5 -Input Type

V0

(1 M

V1

(1 M

V2

(1 M

V3
V4
V5
V6
V7
V8
V9

(1 M
DC
voltage

(1 M
(1 M
(1 M
(1 M
(1 M
(1 M

0 to 10 mV
input resistance)
0 to 50 mV
input resistance)
0 to 60 mV
input resistance)
0 to 100 mV
input resistance)
0 to 1 V
input resistance)
0 to 5 V
input resistance)
1 to 5 V
input resistance)-5 to 5 V
input resistance)
0 to 10 V
input resistance)-10 to 10 V
input resistance)*2: Output
Code
1
2
3
4
5

1.2

Output Type
Current
outputCode

4 to 20 mA
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA

A
B
C
D
E
F
G

Output Type

Voltage
output

How to Read the Model Label
The model label is attached to the right side of the case.

Terminal arrangement
Model (e.g. SGL-A01-0-0)
Input (Enter a type selected in [Input type].)
Output (Enter a type selected in [Output type].)
Serial number(Fig. 1.2-1)- 6 -0 to 10 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V-5 to 5 V

2. Name and Functions
2.1

Front Panel

SGL, SGLL

①

②
③
④
⑤

⑦

⑥

⑧

(Fig. 2.1-1)① Display section② Mounting screw③ DISP key④ MODE key⑤ Up key⑥ DOWN key⑦ Output Zero
⑧ Output Span

Indicates setting contents, input value, output value, etc.
Used for fixing the instrument to the socket or removal from it.
Switches the displays, and moves to the next setting item.
Releases the lock status of the DISP key by pressing for 3 seconds.
Selects either a setting mode or a display mode.
Shifts the digit for the Custom Display.
Enters the setting mode by pressing and holding for 5 seconds.
Increases the numerical value.
Contents of Multi-Display A and B can be changed alternately when
Default Display is RUN display mode 1, 2 and 3.
Decreases the numerical value.
Enters Manual mode by pressing for 3 seconds.
Adjusts the value of Output Zero.
Adjusts the value of Output Span.

- 7 -

2.2

Display Section

SGL, SGLL
①
②
③
④
⑤
⑥
⑦
⑧
⑨
⑩

⑪

⑫
⑬
⑭
⑮

⑯
(Fig. 2.2-1)① Setting display
indicator A② Input indicator A③ Alarm indicator A

Lights up in Manual mode.

Lights up when Multi-Display A indicates an input value.
Lights up if an input error or input disconnection occurs while
Multi-Display A indicates an input value.
Lights up if an input error or input disconnection occurs
in the following mode:
Custom display mode1, 2, 3, Unlit display mode, Model displaymode④ Output indicator A Lights up when Multi-Display A indicates an output value.
⑤ mA indicator
Lights up when mA is selected in [Indication unit].
⑥ % indicator
Lights up in Manual mode or when % is selected in [Indication unit].
⑦ Setting display
Lights up for the setting display.
indicator B⑧ Input indicator B
Lights up when Multi-Display B indicates an input value.
⑨ Alarm indicator B Lights up if an input error or input disconnection occurs while
Multi-Display B indicates an input value.
Lights up if an input error or input disconnection occurs in
Unlit display mode.
⑩ Output indicator B Lights up when Multi-Display B indicates an output value.
⑪ 1 indicator A
Lights up in Manual mode or when Multi-Display A indicates an
output value.
Indicates the following in accordance with the display indication:⑫ Multi-Display A
Input value, output value, custom characters, setting item
⑬ V indicator
Lights up when V is selected in [Indication unit].
⑭ °C indicator
Lights up when °C is selected in [Indication unit].
⑮ 1 indicator B
Lights up when Multi-Display B indicates an output value.
⑯ Multi-Display B
Indicates the following in accordance with the display indication:
Input value, output value, custom characters, setting value
Output indicators A and B, Alarm indicators A and B: Red
Other indicators: White- 8 -3. Mounting
3.1

External Dimensions (Scale: mm)

8P socket

DIN rail(Fig. 3.1-1)

- 9 -

3.2

Mounting to, and Removal from the DIN Rail

Caution• Mount the DIN rail horizontally.
• To remove the socket, a flat blade screwdriver is required.
Never turn the screwdriver when inserting it into the Lock lever. If excessive power
is applied to the lever, it may break.
• If the instrument is mounted in a position susceptible to vibration or shock, mountcommercially available fastening plates at both ends of the instrument.
Recommended Fastening Plate
Manufacturer
Model
End plate
PFP-M
Omron Corporation
Fastening plate BNL6
IDEC Corporation
Fastening plate ATA4806
Panasonic Electric Works Co., Ltd.
Mounting to the DIN rail (Fig. 3.2-1)① Separate the instrument from the socket by loosening the mounting screw on
the front panel.
② Make sure the lock lever of the socket is located in the lower part of the socket.
Hook the upper side of the socket onto the DIN rail, and fit the lower part of thesocket onto the DIN rail (A clicking sound should be heard when done properly).

Caution• Before inserting the instrument to the socket, make sure the cable is wired properly.
(Refer to “4. Wiring”.)• When inserting or removing the socket, make sure the socket is oriented vertically.
If force is applied in any other direction than vertically, a malfunction may occur.
• If the mounting screw is fastened too tightly, a malfunction may occur.
③ Insert the SGL into the socket.
④ Fasten the mounting screw by turning it clockwise, to secure the SGL onto thesocket. Tighten the screw lightly.
Removal from the DIN rail (Fig. 3.2-2)① Turn the power to the instrument OFF.
② Separate the instrument from the socket by loosening the mounting screw on thefront panel.
③ Insert a flat blade screwdriver into the Lock lever (lower part of the socket), andremove the socket from the DIN rail while pulling the lever down.

DIN rail

DIN rail(Fig. 3.2-1)(Fig. 3.2-2)- 10 -

Lock lever4. Wiring
Warning
Turn the power supply to the instrument off before wiring or checking.
Working on or touching the terminal with the power switched on may result insevere injury or death due to electrical shock.
4.1 Lead Wire Solderless Terminal
Use a solderless terminal with an insulation sleeve in which an M3 screw fits as shown
below. The torque should be 0.63 N•m.
Solderless
Manufacturer
Model
Terminal
Nichifu Terminal Industries Co., Ltd.
TMEV1.25Y-3
Y-typeJapan Solderless Terminal MFG Co., Ltd. VD1.25-B3A
Nichifu Terminal Industries Co., Ltd.
TMEV1.25-3
Ring-type
Japan Solderless Terminal MFG Co., Ltd. V1.25-3
Y-type(Scale: mm)

Ring-type(Scale: mm)(Fig. 4.1-1)(Fig. 4.1-2)

4.2 Circuit Configuration
SGL, SGLL
Front
trimmer

A/V
+ ①

Input

④

circuit+Insulation circuit

CPU

Output⑨

circuit⑫

Insulation circuit

⑬

Power

- ⑭

circuit

Display/
Setting
key circuit

Communication
circuit

ModularFor SGLL(Fig. 4.2-1)- 11 -

jack

+
-

4.3

Terminal Arrangement
SGL

SGLL

(Fig. 4.3-1)
PWR
OUT
A
V
RS-485(Fig. 4.3-2)
Power supply 100 to 240 V AC or 24 V AC/DC (for SGL)
OutputDirect current input
DC voltage input
Serial communication (for SGLL)- 12 -

4.4

WiringWarning• For 100 to 240 V AC, if the AC power source is connected to incorrect terminals,
the instrument will be burnt out.
(1) Power Source Wiring
SGL:

Use terminals ⑬, ⑭ for the power supply to the instrument.
For 24 V DC, use terminals ⑬(+), ⑭(-) for the power supply to the
instrument.

SGLL:

Use terminals ⑬, ⑭ for the power supply to the instrument.

(2) Output Wiring
Use terminals ⑨(+), ⑫(-) for the output wiring.
(3) Input Wiring
Use terminals ①, ④ for the input wiring.
(4) Communication Wiring
For the SGLL, connect the SGLL to SGLL using the provided cable.

(Fig. 4.4-1)- 13 -

5. Display Mode
MODE key+
DISP keyfor 3 sec

RUN display
mode 1

MODE key+
DISP keyfor 3 sec

DISP key
RUN display
mode 2

MODE key+
DISP keyfor 3 sec
MODE key+
DISP keyfor 3 sec
MODE key+
DISP keyfor 3 sec

DISP key
RUN display
mode 3
DISP key
Custom display
mode 1DISP key
Custom display
mode 2

MODE key+
DISP keyfor 3 sec
MODE key+
DISP keyfor 3 sec
MODE key+
DISP keyfor 3 sec
MODE key+
DISP keyfor 3 sec

DISP key

Custom display
mode 3
DISP key
Unlit display
mode
DISP key
All unlit display
mode
DISP key
Model display
mode
DISP key
Default Display

Default Display:

RUN display mode 1:
RUN display mode 2:
RUN display mode 3:

If the MODE and DISP keys (in that order) are pressed together
for approx. 3 seconds in any display mode, the display mode will
become the Default Display.
Once the Default Display is set, the DISP key will be in lockstatus.
• If the DISP key is pressed for approx. 3 seconds on the Default
Display, the key lock status will be cancelled.
If the DISP key is pressed while the DISP key is in lock status,
Multi-Display A indicates.
Multi-Display A indicates an input value, and Multi-Display B
indicates the output value.
Multi-Display A indicates an input value, and Multi-Display B
is unlit.
Multi-Display A is unlit, and Multi-Display B indicates the output
value.
The Alarm indicator A will light up if it is under the conditionsof lighting. The Alarm indicator B will light up if it is under theconditions of lighting.
- 14 -Custom display mode 1: Multi-Display A indicates characters set in [Multi-Display A].
Multi-Display B indicates characters set in [Multi-Display B].
The Alarm indicator A will light up if it is under the conditionsof lighting.
Custom display mode 2: Multi-Display A indicates an input value. Multi-Display B
indicates characters set in [Multi-Display B].
The Alarm indicator A will light up if it is under the conditionsof lighting.
Custom display mode 3: Multi-Display A indicates an output value. Multi-Display B
indicates characters set in [Multi-Display B].
The Alarm indicator A will light up if it is under the conditionsof lighting.
Unlit display mode:
Multi-Display A and B are unlit, and the Input indicator A lights
up.
The Alarm indicator A will light up if it is under the conditionsof lighting.
All unlit display mode:
All displays and indicators are unlit.
Alarm indicator A and B do not light up even if they are under theconditions of lighting.
Model display mode:
Multi-Display A indicates a model name, and Multi-Display B
indicates an input and output code.
The Alarm indicator A will light up if it is under the conditionsof lighting.

- 15 -

6. Setting Mode
6.1

Display Transition in Setting Mode•

Available only for the SGLL.
• If the MODE key is pressed and held down for approx. 5 seconds in each setting
mode, the unit will move to the Default Display.
Default Display
DOWN
key
for 3 sec

MODE
key
for 5 sec

Manual mode
(See p.35.)
MODE key, or
after Manualmode auto
return time has
elapsed.
Returns to theDefault Display.

Input setting
mode(See pp.18, 19.)
DISP
key
for 3 sec
Input type

MODE
key

Outputsetting mode
(See pp.20, 21.)
DISP
key
for 3 sec
Output type
DISP
key

DISP
key
Input decimal
point place

Output decimal
point place

DISP
key

DISP
key
Output 0% value

Indication value
at output 0%
DISP
key

DISP
key

Indication value
at output 100%

MODE
key

Linearization
setting mode
(See pp.22-30.)
DISP
key
for 3 sec
Linearization
Enabled/Disabled
DISP
key
Input X1

DISP
key
Output Y1
DISP
key
Input X2

DISP
key

DISP
key

DISP
key

Indication unit

Save settings

Output Y24

DISP
key

DISP
key

DISP
key

Save settings

Returns to theDefault Display.

Input X25
DISP
key

DISP
key
Returns to theDefault Display.

Output Y25
DISP
key
Save settings
DISP
key
Returns to theDefault Display.

- 16 -

MODE
key

Instrument
setting mode
(See pp.31, 32.)
DISP
key
for 3 sec

DISP
key
Sensor correction

MODE
key

Communication
setting mode
(See pp.33, 34.)

MODE
key

Custom display
setting mode
(See p.35.)

DISP
key
for 3 sec

DISP
key
for 3 sec

Instrument
numberMulti-display A

DISP
key

DISP
key

Communication
speed

Multi-display B

DISP
key

DISP
key

DISP
key

Indication time

Data bit/Parity

Save settings

DISP
key

DISP
key

DISP
key

Auto/Manual
DISP
key
Manual mode
auto return time
DISP
key
Save settings
DISP
key
Returns to theDefault Display.

Stop bit

Returns to theDefault Display.

DISP
key
Response delay
time
DISP
key
Save settings

DISP
key
Returns to theDefault Display.

- 17 -MODE
key

To Input setting mode

6.2

Input Setting Mode

Input type
Selects an input type.
Setting Range

Indication
Multi-Display A Multi-Display B

4 to 20 mA
Built-in 50
shunt resistor
4 to 20 mA
Externally mounted 250
shunt resistor
4 to 20 mA
Externally mounted 50
shunt resistor
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA
1 to 5 mA
0 to 1 mA
10 to 50 mA
0 to 10 mV
0 to 50 mV
0 to 60 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V-5 to 5 V
0 to 10 V-10 to 10 V

Factory
Default

4 to 20 mA
Built-in 50
shunt resistor

Input Decimal Point Place
Selects the decimal point place for input value indication.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
No decimal point
2 digits after
1 digit after decimal point
decimal point *2 digits after decimal point
3 digits after decimal point* Factory default will be ‘1 digit after decimal point’ if the following input types are
selected:
0 to 10 mV, 0 to 50 mV, 0 to 60 mV, 0 to 100 mV, -5 to 5 V, -10 to 10 V

- 18 -

Output 0% Value
Sets an input value (indicated on the display) at the time of output 0%.
Values change in accordance with the input type.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
4.00 *
Low limit of each input type to
Set value
Output 100% value* If [-5 to 5 V] is selected in [Input type], the factory default value will be -50.0.
Output 100% Value
Sets an input value (indicated on the display) at the time of output 100%.
Values change in accordance with the input type.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
20.00 *
Output 0% value to
Set valueHigh limit of each input type
* If ‘-5 to 5 V’ is selected in [Input type], the factory default value will be 50.0.

入Indication Unit
Selects the unit for indication.
Setting Range

Indication
Factory Default
Multi-Display A Multi-Display B

No unit%mA
V°C

No unit

Save Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Save
Save
Not save- 19 -

6.3

Output Setting Mode

Output Type
Selects an output type.
Setting Range

Indication
Factory Default
Multi-Display A Multi-Display B

4 to 20 mA
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA
0 to 10 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V-5 to 5 V

4 to 20 mA

Output Decimal Point Place
Selects a decimal point place for output value indication.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
No decimal point
2 digits after
1 digit after decimal point
decimal point *2 digits after decimal point
3 digits after decimal point* The factory default value will be ‘1 digit after decimal point’ when the following type isselected in [Output type]:
0 to 10 mV, 0 to 100 mV, -5 to 5 V
Indication Value at Output 0%
Sets an indication value at the time of output 0%.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
4.00-1999 toSet value
Indication value at output 100%

- 20 -

Indication Value at Output 100%
Sets an indication value at the time of output 100%.
Setting Range

Indication
Multi-Display A Multi-Display B

Factory Default
20.00

Indication value at output 0% to9999

Set value

Save Settings
Selects whether the settings are saved (registered) or not
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Save
Save
Not save- 21 -

6.4

Linearization Setting Mode

Linearization Enabled/Disabled
Selects the Linearization function Enabled or Disabled.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Disabled
EnabledDisabled
Input X1
Sets Input X1 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y1
Sets Output Y1 corresponding to Input X1 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value
Input X2
Sets Input X2 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y2
Sets Output Y2 corresponding to Input X2 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 22 -

Input X3
Sets Input X3 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y3
Sets Output Y3 corresponding to Input X3 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X4
Sets Input X4 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y4
Sets Output Y4 corresponding to Input X4 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X5
Sets Input X5 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y5
Sets Output Y5 corresponding to Input X5 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 23 -

Input X6
Sets Input X6 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y6
Sets Output Y6 corresponding to Input X6 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X7
Sets Input X7 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y7
Sets Output Y7 corresponding to Input X7 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X8
Sets Input X8 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y8
Sets Output Y8 corresponding to Input X8 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 24 -

Input X9
Sets Input X9 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y9
Sets Output Y9 corresponding to Input X9 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X10
Sets Input X10 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y10
Sets Output Y10 corresponding to Input X10 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X11
Sets Input X11 for linearization.。
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y11
Sets Output Y11 corresponding to Input X11 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 25 -

Input X12
Sets Input X12 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y12
Sets Output Y12 corresponding to Input X12 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X13
Sets Input X13 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y13
Sets Output Y13 corresponding to Input X13 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X14
Sets Input X14 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y14
Sets Output Y14 corresponding to Input X14 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 26 -

Input X15
Sets Input X15 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y15
Sets Output Y15 corresponding to Input X15 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X16
Sets Input X16 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y16
Sets Output Y16 corresponding to Input X16 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X17
Sets Input X17 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y17
Sets Output Y17 corresponding to Input X17 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 27 -

Input X18
Sets Input X18 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y18
Sets Output Y18 corresponding to Input X18 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X19
Sets Input X19 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y19
Sets Output Y19 corresponding to Input X19 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X20
Sets Input X20 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y20
Sets Output Y20 corresponding to Input X20 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 28 -

Input X21
Sets Input X21 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y21
Sets Output Y21 corresponding to Input X21 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X22
Sets Input X22 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y22
Sets Output Y22 corresponding to Input X22 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X23
Sets Input X23 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y23
Sets Output Y23 corresponding to Input X23 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value- 29 -

Input X24
Sets Input X24 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y24
Sets Output Y24 corresponding to Input X24 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Input X25
Sets Input X25 for linearization.
Setting Range
Output 0% value to
Output 100% value

Indication
Factory Default
Multi-Display A Multi-Display B
Output 0% value
Set value
Set value

Output Y25
Sets Output Y25 corresponding to Input X25 for linearization.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0.0%
0.0 to 100.0%
Set value

Save Settings
Selects whether the settings are saved (registered) or not
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Save
Save
Not save- 30 -

6.5

Instrument Setting Mode

Filter Time ConstantSets the input filter time constant.
Input fluctuation due to noise can be decreased.
Setting Range

Indication
Multi-Display A Multi-Display B

Factory Default
0.0 seconds

0.0 to 10.0 seconds

Set valueSensor Correction
Sets the sensor correction value.
Input value = Current input value + (Sensor correction value)
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
0-1000 to 1000 *Set value* The placement of the decimal point follows the selection.
Indication Time
Sets duration from no operation until indication (of Multi-Display A, Multi-Display B,
and each action indicator) turns off.
They remain lit during setting mode or in the event of an input error or input
disconnection.
When set to 00.00, they remain lit.
After indication time has elapsed, if any key is pressed while they are unlit,
indication will light up again.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
30 : 00
00 : 00 to 60 : 00 (Minutes : Seconds)(Minutes :00 : 00 ................. Continuous
Set value
Seconds)
00 : 01 to 60 : 00 . Indication time

Auto/Manual
If AUTO is selected, the output value will be output corresponding to the input value.
When MANUAL is selected, the unit can enter Manual mode. The output value set in
Manual mode will be output.
Indication
Factory Default
Setting Range
Multi-Display A Multi-Display B
ManualAuto
Manual- 31 -

Manual Mode Auto Return Time
Sets duration from manual mode until the unit automatically returns to the Default
Display.
If set to 0 (zero), auto return will not occur.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
30 minutes
0 to 60 minutes

Set value

Save Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Save
Save
Not save- 32 -

6.6 Communication Setting Mode
Available only for the SGLL.
Instrument Number
Sets an instrument number.
Setting Range

Indication

Factory Default

Multi-Display A Multi-Display B

1
1 to 247

Set value

Communication Speed
Selects the communication speed.
Setting Range

Indication
Multi-Display A Multi-Display B

9600 bps
19200 bps
38400 bps

Factory Default
38400 bps

Data bit/Parity
Selects data bit and parity.
Setting Range

Indication
Multi-Display A Multi-Display B

8 bits/No parity
8 bits/Even
8 bits/Odd

Factory Default
8 bits/Odd

Stop Bit
Selects the stop bit.
Setting Range

Indication

Factory Default

Multi-Display A Multi-Display B

1 bit

1 bit
2 bitsResponse Delay Time
Response from the instrument can be delayed after receiving command from the hostcomputer.
Setting Range

Indication
Multi-Display A Multi-Display B

Factory Default
10 ms

0 to 1000 ms

Set value- 33 -

Save Settings
Selects whether the settings are saved (registered) or not
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Save
Save
Not save- 34 -

6.7 Custom Display Setting Mode
Customizes characters to be indicated on the Multi-Display A and B*.
Use alphanumeric characters and symbols.
(e.g.) FLOW, TEMP, No.1, No.2
* Number of characters which can be indicated differs depending on the display
mode.
Refer to Section ‘エラー! 参照元が見つかりません。’ (pp.14, 15).
• If Custom display mode 1 is selected:
Up to 8 characters can be displayed in total for both Multi-Display A and B.
• If any of Custom display mode 2 to 3 is selected:
Up to 4 characters can be displayed on the Multi-Display B.
Can be set from the thousands digit of the display.
Digits can be selected with the MODE key.
Multi-Display A
Characters for the Multi-Display A can be customized.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
AAAA
A to Z, 0 to 9, /, –，. , (Blank)
Set value

Multi-Display B
Characters for the Multi-Display B can be customized.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
AAAA
A to Z, 0 to 9, /, –，. , (Blank)
Set value
Save Settings
Selects whether the settings are saved (registered) or not
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B
Save
Save
Not save
6.8

Manual Mode
If MANUAL is selected in [Auto/Manual] in Instrument setting mode, press the DOWN
key on the Default Display for approx. 3 seconds. The unit will enter Manual mode.
At this time, Multi-Display A flashes the output value.
The output value can be set by the UP or DOWN key.
The output value will be lit while setting.
By pressing the MODE key in Manual mode, or after Manual mode auto return
time has elapsed, the unit returns to the Default Display, and outputs the value
corresponding to the input value.
- 35 -

7. Adjustment
Performs the output zero and span adjustments.
For this instrument, the output adjustment has already been completed when shipped.
If the instrument is used with the ordered Input/Output spec, the adjustment is not required.
However, for calibration, or for the fine adjustment of the SGL to which any equipment isconnected, perform the adjustment.
Connect a voltage current generator to the input terminals of this instrument.
Connect a digital multimeter to the output terminals.
7.1

Basic Operation of Adjustment
For adjustment, use the following trimmers on the front panel.
Output Zero: Adjusts the value of Output Zero.
Output Span: Adjusts the value of Output Span.

7.2

Adjustment
Perform adjustment as follows.

7.2.1 Output Adjustment
The following outlines the procedure for Output adjustment.
① Enter the value corresponding to output 0%, and adjust the value using the‘Output Zero’ trimmer while viewing the output value (on the digital multimeter).
② Enter the value corresponding to output 100%, and adjust the value using the‘Output Span’ trimmer while viewing the output value (on the digital multimeter).
③ Enter the value corresponding to output 0% again, and confirm the output value(on the digital multimeter).
④ If the value corresponding to output 0% is not at 0%, repeat steps ① to ③
again.

- 36 -

8. Operation
8.1

Indication after Power-on

After the power is turned on, the instrument is switched to warm-up status for 3
seconds. Multi-Display A indicates a model name, and Multi-Display B indicates the
input code and output code.
(e.g.) SGL-A01-0-0
Multi-Display A:
Multi-Display B:
For the output, the output value corresponding to input 0% will be output.
8.2 Operation
After warm-up status, the unit enters display mode.
The input signal selected in [Input type] will be converted to the output selected in[Output type].
8.2.1 Input Indication Range
The input value is indicated within the following range:[Input range low limit + Input span x 10%] to[Input range high limit + Input span x 10%]
For a value lower than (and including) -2000, the minus (-) sign and the input value
are indicated alternately. For a value higher than (and including) 10000, the lower 4
digits will flash. (The placement of the decimal point follows the selection.)
When exceeding the indication range,
will flash.
When dropping below the indication range,
will flash.
8.2.2 Output Indication Range
The output value is indicated within the following range:[Indication value at output 0% – (Indication value at output 100% – Indication
value at output 0%) ×10%] to[Indication value at output 100% + (Indication value at output 100% – Indication
value at output 0%) ×10%]
For a value lower than (and including) -2000, the minus (-) sign and the output value
are indicated alternately. For a value higher than (and including) 10000, the lower 4
digits will flash. (The placement of the decimal point follows the selection.)- 37 -

8.2.3 Input Disconnection Status
If input is disconnected, the input status will become as follows.
Input Range
Input Status
4 to 20 mA
Equals 0 mA input.
Built-in 50
shunt resistor
4 to 20 mA
Externally mounted 250
shunt
Equals 0 mA input.
resistor
4 to 20 mA
Externally mounted 50
shunt
Equals 0 mA input.
resistor
0 to 20 mA
Equals 0 mA input.
0 to 16 mA
Equals 0 mA input.
2 to 10 mA
Equals 0 mA input.
0 to 10 mA
Equals 0 mA input.
1 to 5 mA
Equals 0 mA input.
0 to 1 mA
Equals 0 mA input.
10 to 50 mA
Equals 0 mA input.
0 to 10 mV
Overscale *0 to 50 mV
Overscale *0 to 60 mV
Overscale *0 to 100 mV
Overscale *0 to 1 V
Overscale *0 to 5 V
Equals 0 V input.
1 to 5 V
Equals 0 V input.
-5 to 5 V
Equals 0 V input.
0 to 10 V
Equals 0 V input.
-10 to 10 V
Equals 0 V input.
* For the overscale status, the Alarm indicator lights up, andas an input value.

flashes

8.2.4 Indication Time Setting
After preset indication time has elapsed, Multi-Display A, Multi-Display B andeach action indicator are turned OFF.
They will light up again if any key is pressed.
They remain lit during setting mode or in the event of an input error or input
disconnection.
If the indication time is set to 00:00, they will remain lit.

- 38 -

8.2.5 Linearization Function
By setting up to 25 points of input and output, the ratio of input and output can bechanged.
One point consists of one pair of input and output.
Plots points starting from the smallest input value (X1, Y1), (X2, Y2) … (Xn, Yn) in
numerical order.
(n: Numbers from 1 to 25)
However, if Disabled is selected in [Linearization Enabled/Disabled], the linearization
function will be disabled.
Output value
110%
100%
Curved line
characteristics

Y3

Linearcharacteristics

Output (Yn)
Y2
Y1
0%
①
-10%

②

X1

③

④

X2 X3

Output 0% value

Output 100% value

Input (Xn)
(1) If input is lower than X1
Outputs linearly between -10% and Y1.
(2) In the case of X1 Input value < X2
Outputs linearly between Y1 Output value<Y2.
(3) In the case of X2 Input value<X3
Outputs linearly between Y2 Output value<Y3.
(4) For other inputs which follow the above, outputs linearly under the same
conditions as steps (2) and (3), depending on points.
Outputs linearly between the last Xn and 110%.
If Xn and Xn+1 are set to the same value under the above conditions, this isconsidered as no setting, and for settings after Xn+1, the linearization function will
be invalidated.
If the same values exist from X1 to Xn, only the smallest “n” will be effective.
- 39 -

9. Specifications
Input Specifications
Direct current
input

DC voltage input

Shunt Resistor
50 *4 to 20 mA DC
250
50
0 to 20 mA DC
250
0 to 16 mA DC
62.5
2 to 10 mA DC
250
0 to 10 mA DC
100
1 to 5 mA DC
100
0 to 1 mA DC
1000
10 to 50 mA DC
10* Built-in shunt resistor

Indication Resolution
1
1
1
1
1
1
1
1
1
1

Shunt Resistor

Indication Resolution
1
1
1
1
1
1
1
1
1
1

Input Range

Input Range
0 to 10 mV
0 to 50 mV
0 to 60 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V-5 to 5 V
0 to 10 V-10 to 10 V

1M

Output Specifications
Direct current

Allowable
Zero
Load
Adjustment
ResistanceRange
4 to 20 mA
750
max.
0 to 20 mA *750
max.
0 to 16 mA *900
max.
-5 to 5%2 to 10 mA
1500
max.
0 to 10 mA * 1500max.
* 0 mA or less: Out of base accuracy
Output Range

DC voltage

Allowable
Zero
Load
Adjustment
ResistanceRange
0 to 10 mV *10 k min.
0 to 100 mV *100 k min.
0 to 1 V * 1000min.
0 to 5 V * 5000
min.
-5 to 5%
1 to 5 V
5000
min.
0 to 10 V *10 k min.
-5 to 5 V
10 k min.
* 0 V or less: Out of base accuracy
Output Range- 40 -

Span
Adjustment
Range
95 to 105%

Span
Adjustment
Range

95 to 105%

Performance
Base accuracy(at 25°C)

0.1% of each input span

Temperature
coefficient

0.015 %/°C
0 to 10 mV output:

Response time

500 ms max. (0→90%)

Indication update
cycle

125 msIndication
accuracy

Base accuracy 1 digit

Insulation
resistance

100 M

0.02 %/°C

minimum, at 500 V DC

Dielectric strength 2.0 kV AC for 1 minute
General Structure
Dimensions

22.5 x 89 x 70 mm (W x H x D)

WeightApprox. 77 g

Mounting method

DIN railCase

Flame-resistant resin, Color: Black

Front panel

Polycarbonate

Installation Specifications
Power supply

100 to 240 V AC 50/60 Hz
24 V AC/DC 50/60 Hz

Allowable voltage
range

100 to 240 V AC: 85 to 264 V AC
24 V AC/DC: 20 to 28 V AC/DC

Power
consumption

100 to 240 V AC: Approx. 9 VA max. (SGLL: Approx. 10 VA max.)
24 V AC: Approx. 6 VA max.
24 V DC: Approx. 3 W max.

Ambient temperature -10 to 55°C (Non-condensing, no icing)Ambient humidity

35 to 85 %RH (Non-condensing)- 41 -

Serial Communication (for SGLL)
Operation from an Reading and setting of various set values
external computer Reading of the input value and action status
Function change
Communicationline

EIA RS-485

Communication
methodHalf-duplex communication

Communication
speed

9600, 19200, 38400 bps (Selectable by keypad)(Factory default: 38400 bps)

Synchronization
methodStart-stop synchronization

Communication
protocolModbus RTU

Start bit

1 bit

Data bit

8 bitsParityEven/Odd/No parity (Selectable by keypad)(Factory default: Odd)

Stop bit

1 bit or 2 bits (Selectable by keypad)(Factory default: 1 bit)

Response delaytime

Response from the instrument can be delayed after receiving
command from the host computer.
0 to 1000 ms (Factory default: 10 ms)

Standard Function
Power failurecountermeasure

The setting data is backed up in the non-volatile IC memory.

Self-diagnosis

The CPU is monitored by a watchdog timer, and if an abnormal
status occurs, the instrument is switched to warm-up status, turning
all outputs OFF.

- 42 -

10. Troubleshooting
10.1 Indication
Problem
Multi-Display A or B
flashesorwhen itindicates an input value.

Multi-Display A or B isirregular or unstable
when it indicates an input
value.

Displays and indicators
are unlit.
If any key is pressed,
they will light up.
10.2 Key Operation
ProblemIf the DISP key ispressed, Multi-Display A
shows, and thedisplay mode cannot beswitched.
10.3 Operation
Problem
When Multi-Display A or
B indicates an input
value, the input value
does not change.

No output

Possible Cause
The sensor may bedisconnected.
Check whether the sensor
is securely mounted to the
input terminals of this
instrument.
Check the input signal
source.
Check whether sensor input
is correct.
Sensor correction value is
unsuitable.
AC leaks into the sensor
circuit.
There may be equipmentthat interferes with or makes
noise near the instrument.
The Indication Time (p.31) is
set to any value other than
00 : 00.
(Factory default is 30 : 00.)

Solution
Replace with a new sensor.
Connect the sensor
terminals to the instrument
input terminals securely.
Ensure that the input signal
source works normally.
Select the same sensor
type as that of currently used
sensor.
Set it to a suitable value.
Use an ungrounded typesensor.
Keep the instrument clear ofany potentially disruptive
equipment.
To indicate continuously,
set the Indication Time (p.31)
to “00 : 00”.

Possible Cause
The DISP key is in locked
status.

Solution
Press the DISP key for
approx. 3 seconds to release
the key lock.

Possible Cause
The sensor may be out oforder.
Check whether input andoutput wires are securely
connected to the I/O
terminals of the instrument.
Check whether the wiring of
input and output are correct.
Selections in [Output type(p.20)] may be incorrect.

Solution
Replace with the newsensor.
Ensure that input and output
wires are securely
connected to the I/O
terminals of the instrument.
Wire them correctly.

- 43 -

Make a correct selection in[Output type (p.20)].

11. Character Table
Please use the following factory default values for your reference.
Display mode
Setting Item
Default display mode
RUN display mode 1
RUN display mode 2
RUN display mode 3
Custom display mode 1
Custom display mode 2
Custom display mode 3
Unlit display modeAll unlit display mode
Model display mode
Setting mode
Setting Item
Input setting mode
Output setting mode
Linearization setting mode
Instrument setting mode
Communication setting mode
Custom display setting mode
Input setting mode
Setting Item
Input type
Input decimal point place
Output 0% value
Output 100% value
Indication unit
Save settings
Output setting mode
Setting Item
Output typeOutput decimal point place
Indication at output 0%
Indication at output 100%
Save settings
Linearization setting mode
Setting Item
Linearization
Enabled/Disabled
Input X1

Multi-Display A
Multi-Display B
Follows currently indicated display mode.
Input value
Output value
Input value
Unlit
Unlit
Output value
Input value
Output value
Unlit(Input indicator A lit)
Unlit
Model

Data

Unlit
Unlit
Input, Output codes

Multi-Display A

Multi-Display B
Unlit
Unlit
Unlit
Unlit
Unlit
Unlit

Data

Multi-Display A

Multi-Display B

Data

Multi-Display A

Multi-Display B

Data

Multi-Display A

Multi-Display B

Data

Output 0% value- 44 -

Setting Item
Output Y1
Input X2
Output Y2
Input X3
Output Y3
Input X4
Output Y4
Input X5
Output Y5
Input X6
Output Y6
Input X7
Output Y7
Input X8
Output Y8
Input X9
Output Y9
Input X10
Output Y10
Input X11
Output Y11
Input X12
Output Y12
Input X13
Output Y13
Input X14
Output Y14
Input X15
Output Y15
Input X16
Output Y16
Input X17
Output Y17
Input X18
Output Y18
Input X19
Output Y19
Input X20
Output Y20
Input X21
Output Y21
Input X22
Output Y22
Input X23

Multi-Display A

Multi-Display B
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value
Output 0% value- 45 -

Data

Setting Item
Output Y23
Input X24
Output Y24
Input X25
Output Y25
Save settings
Instrument setting mode
Setting Item
Filter time constantSensor correction
Indication time
Auto/Manual
Manual mode auto return time
Save settings

Multi-Display A

Data

Output 0% value
Output 0% value

Multi-Display A

Communication setting mode (for SGLL)
Setting Item
Multi-Display A
Instrument number
Communication speed
Data bit/Parity
Stop bit
Response delay time
Save settings
Custom display setting mode
Setting Item
Multi-Display A
Multi-Display B
Save settings

Multi-Display B

Multi-Display A- 46 -

Multi-Display B

Data

Multi-Display B

Data

Multi-Display B

Data- 47 -

***** Inquiries *****For any inquiries about this unit, please contact our agency or thevendor where you purchased the unit after checking the following.
[Example]• Model ------------------------------ SGL-A01-0-0• Serial number -------------------- 154F05000
In addition to the above, please let us know the details of themalfunction, or discrepancy, and the operating conditions.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, Japan[URL]
https://shinko-technos.co.jp/e/Tel: +81-72-727-6100[E-mail]overseas@shinko-technos.co.jp
Fax: +81-72-727-7006

No.SGL11E5

2023.04