Pressure transmitter

Pressure transmitter for
refrigeration applications
Type 560
Huba Control is expanding its product family of pressuretransmitters with a new product series for OEM customers in
the refrigeration sector.
The 560 features the well known Huba Control stainless steel
sensor technology that has proven itself millions of times
over. Thanks to the hermetically welded construction, the560 is suitable for known refrigerants such as ammonia andCO2 as well as other refrigerants from safety classes A2L andA3. This gives you great flexibility within your applications.
With a wide range of variants, such as the copper soldering
tube, the pressure transmitter type 560 can be optimally
integrated into your application.

Pressure range-1 ... 7 bar
0 ... 10 – 150 bar+ Compact and robust design+ Hermetically sealed in accordance with EN ISO14903:2017
tested during manufacture by means of 100% helium 		leakage control+ Due to the hermetically welded construction, no elastomer
seals are required+ Two colour versions of connectors for easy identificationof pressure ranges within an application+ Rated according to DIN EN 60335-2-40+ Copper soldering tube for optimal integration into yourcopper pipe system

Pressure and flow

Huba Control Type 560 | Technical data subject to change | Edition 01/2024

1/6

Technical overview
Pressure range
Relative					-1 ... 7 bar / 0 ... 10 – 150 bar

Operating conditions
Medium								Mediumratiom. 10 ... 90%			
Medium4 ... 20 mA
Temperature			
Ambient					Storage		
Tolerable overload				Rupture pressure				Materials
Cover					Plug accommodation								Thread		Materials in contact with medium 3)			
Soldering connection			Sensor		
Electrical overview
2 wire3 wirePolarity reversal protection
Insulation voltage

Refrigerants-40 ... +125 ºC-40 ... +120 ºC-40 ... +85 ºC-40 ... +100 ºC
3 x fs 1)
6 x fs 2)

Stainless steel 1.4301 / 1.4404
Polyacrylamide 50% GF UL94 V0
Stainless steel 1.4301 / AISI 304
CuDHP / CW024A
Stainless steel 1.4016

OutputPower supply
Load		Current consumption
voltage - 8 V< supply0.02
4 ... 20 mA
8 ... 32 VDC< 23 mA
Aratiom. 10 ... 90%
2.97 ... 5.5 VDC>10 kOhm / < 10 nF< 7 mA
Short circuit proof and protected against polarity reversal. Each connection is protected against crossover up to max. supply voltage.
500 VDC

Dynamic response
Response time						ratiom. 10 ... 90%
4 ... 20 mA		

< 5 ms (typ. 2 ms)< 5 ms (typ. 2 ms)

Electrical connection 			Metri Pack 150 P2S Series 4)(two coloured design, black or grey)
Swift connector(two coloured design, black or grey)

Protection standard
IP 67 5)		IP 67 5)		Protection class
III
III

Pressure connection
Inside thread/16 - 20 UNF/16 - 20 UNF¼ -18 NPTG¼
G¼
Tube Ø 6 mm

sealing cone 45º and Schrader
sealing cone 45ºsealing in thread

77

Outside thread
Soldering connection 6)(other temperatures on request)

sealed at back DIN EN ISO 1179-2 with profile seal ring in FKM
sealed at back DIN EN ISO 1179-2 with profile seal ring in EDPM

L = 35 mm

Installation arrangement
Unrestricted
Tests / Admissions
UL		
Electromagnetic compatibility		
Hermetic sealing		Surface temperature during operation and individual faults

ANSI/UL 61010-1 acc. E325110
CE-Conform acc. EN 61326-2-3:2013
Tested according to EN ISO 14903:2017
Assessment according to DIN EN 60335-2-40

Weight~ 90 g

Packaging (recycleable)

Multiple packaging in cardboard (25 pcs.) 					min. order quantity 100 pcs.

Accuracy
Parameter

Unittypic.

max.

Characteristic line acc. IEC 61298-2 7)

% FS

0.5

1Resolution

% FS

0.1

Thermal characteristic -20 ... +85 °C

% FS

1

1.5

Thermal characteristic -40 ... +125 °C

% FS

1.5

2

Long term stability acc. IEC 61298-2

% FS

0.2

Soldering tube with pressure range ≥90 bar: max. 200 bar4)with media contact.
Delivery without female connector.
approval
1)

2/6

Pressure and flow

Test conditions: 25 °C, 45% rF3)
Soldering tube with pressure range ≥90 bar: max. 600 bar
The application medium must be suitable for the materials of the pressure transmitter
The specified protection class only applies when connected with a matching plug of the corresponding protection class; IP test not part of UL61010-1
6)
7)
Soldering tube with end stop, without end stop on request.
incl. zero point, full scale, linearity, hysteresis and repeatability2)
5)

Huba Control Type 560 | Technical data subject to change | Edition 01/2024

1

Order code selection table in bar

560.

234567891011X X X X X X X X X X X-1 ... 7 bar					9 1 1 		 0, 50 ... 10 bar					9 1 2		 0, 50 ... 16 bar					9 1 3		 0, 50 ... 18 bar					9 1 4		 0, 50 ... 30 bar					9 1 5		Pressure range (relative)
0 ... 45 bar					9 1 6		0 ... 50 bar					9 1 7			0 ... 60 bar					9 1 8			0 ... 90 bar					9 1 9		0 ... 150 bar					9 1 A		Ratiom. 10 ... 90%		
2.97 ... 5.5 VDC
3 wire				7Output / power supply
Analog 4 ... 20 mA
8 ... 32 VDC
2 wire				3
Swift connector (black)		
2L: IN=1 / OUT=3
3L: IN=1 / OUT=2 / GND=3					0Swift connector (grey)		
2L: IN=1 / OUT=3
3L: IN=1 / OUT=2 / GND=3					AElectrical connection1)		
Metri Pack 150 P2S (black)
2L: IN=B / OUT=A
3L: IN=B / OUT=C / GND=A					5		Metri Pack 150 P2S (grey) 1)
2L: IN=B / OUT=A
3L: IN=B / OUT=C / GND=A					
B7Inside thread		/16 -20 UNF with Schrader							K 00 007/16 -20 UNF								200 00			¼ -18 NPT								 Y00 00Pressure connection
Outside thread			G ¼ sealed at back DIN EN ISO 1179-2 with profile seal ring in FKM						 4 A0 00			G ¼ sealed at back DIN EN ISO 1179-2 with profile seal ring in EPDM 						4 B0 00
Soldering connection
Tube Ø 6mm 		
L = 35 mm						 W 00 00
Pressure range variation
Indicate W and state range on order (e.g. W -1 ... +3 bar/Out 10 ... 90%)												W(optional)

1

Order code selection table in psi

560.

2

34567891011X X X X X X X X X X X-15 ... 100 psi					9 A 1 		 0, 50 ... 145 psi					9 A 2		 0, 50 ... 230 psi					9 A 3		 0, 50 ... 260 psi					9 A 4		 0, 50 ... 435 psi					9 A 5		Pressure range (relative)
0 ... 650 psi					9 A 6		0 ... 725 psi					9 A 7			0 ... 870 psi					9 A 8			0 ... 1300 psi					9 A 9		0 ... 2170 psi					9 A A		Ratiom. 10 ... 90%		
2.97 ... 5.5 VDC
3 wire				7Output / power supply
Analog 4 ... 20 mA
8 ... 32 VDC
2 wire				3
Swift connector (black) 		2L: IN=1 / OUT=3
3L: IN=1 / OUT=2 / GND=3					0Swift connector (grey) 		2L: IN=1 / OUT=3
3L: IN=1 / OUT=2 / GND=3					AElectrical connection1)		
Metri Pack 150 P2S (black)
2L: IN=B / OUT=A
3L: IN=B / OUT=C / GND=A					5		Metri Pack 150 P2S (grey) 1)
2L: IN=B / OUT=A
3L: IN=B / OUT=C / GND=A					B7Inside thread		/16 -20 UNF with Schrader							K 00 00
7/16 -20 UNF								200 00			¼ -18 NPT								 Y00 00
Outside thread
Pressure connection			G ¼ sealed at back DIN EN ISO 1179-2 with profile seal ring in FKM						 4 A0 00			G ¼ sealed at back DIN EN ISO 1179-2 with profile seal ring in EPDM 						4 B0 00Soldering connection
Tube Ø 6mm 		
L = 35 mm						 W 00 00Pressure range variation
Indicate W and state range on order (e.g. W -15 ... +45 psi/Out 10 ... 90%)												W(optional)

Variation
Variations
Deviation factor

0.33 ... 1.33 (must not be undercut or exceeded)

AccuracyStandard accuracy / variation factor (for variation factor < 1)

Lowest starting point (relative)-1 bar

Ex. 1: Sensor with a measuring range of 0 ... 10 bar is to be calibrated to 0 ... 4 bar.								4 barDeviation factor == 0.4  ok, because the deviation factor is within 0.33 ... 1.33.
10 bar
Ex. 2: Sensor with a measuring range of 0 ... 10 bar is to be calibrated to 0 ... 2 bar.								2 barDeviation factor == 0.2  cannot be synchronized, as the deviation factor is not within 0.33 ... 1.33.
10 bar1)

Delivery without female connector

Pressure and flow

Huba Control Type 560 | Technical data subject to change | Edition 01/2024

3/6

Accessories					Order number
Set - Swift connector (female) (included in delivery)			
Set - Metri Pack 150 P2S Series (female) (3 wire)			

117312120345

Mounting instruction soldering

Sheath temperature:max. 125 °C

Surface temperature of the sensor in the event of a fault as a function of the supply power. Measured according to EN60335-2-40

P [W]

Ratiometer 10 ... 90%

Surface temperature [°C]

P [W]

Analog 4 ... 20 mA

Surface temperature [°C]Electrical connections
Swift connector
2 wirebraun brown/ brown / brun

white
weiss
/ white / blancgrün green/ green / vert

4/6

Metri Pack Series 150
3 wire3 wirewhite
weiss / white/ blanc

grün green/ green / vert

11(IN)(IN) 33(OUT)(OUT)(IN) 22 (OUT)
1 (IN)(OUT) 33(GND)(GND)

B (IN) A (OUT)

B (IN) C (OUT) A (GND)

Pressure and flow

2 wirebraun brown/ brown / brun

B B(IN)(OUT)(IN)A A
(OUT)(IN) CC(OUT)(OUT) AA(GND)(GND)
BB (IN)

Huba Control Type 560 | Technical data subject to change | Edition 01/2024

Dimensions in mm / Electrical Connections

100 (Referenzmass für Massstab)≈28,32410Nmn18,8

7/16-20 UNF
inside with Schrader

66
≈28,3

12,72420Nm

Metri Pack 150 P2S Series≈46,6

Swift Connectorn24

1/4"-18 NPT≈47,8n24

12,3

612G1/4"
DIN 3852-E

2420Nm

6167/16-20 UNF

2410Nm45°

Dieses Dokument ist Eigentum der Huba Control AG, welcher alle Rechte
an diesem Dokument vorbehalten sind. Die Zeichnung darf weder an
Dritte weitergegeben, vervielfältigt oder anderweitig verwendet werden.
Es sei denn mit schriftlicher Zustimmung der Huba Control AG.

666,7n635Tolerierungsgrundsatz: ISO 8015 Allgemeintoleranzen: ISO 22081¬=Prüfmerkmal, XXX =Besonderes Merkmal, u =Zulassungsmerkmal
Artikelgruppe:alte Zeichn. Nr.:

Unterliegt folgenden Zulassungen:.

Zeichnungs-Nr.:

Lineare Grössenmasse: Toleranzklasse JS13/js13, siehe
ISO 286-1
RESTRICTED Status:
Format:A3

Massstab:

1:1

Assembly_Flyer_DB_EN
Pressure and flow

Gezeichnet:
Geändert:

29.09.2023.

Index:

Ulli
Seite:

2/2

Huba Control Type 560 | Technical data subject to change | Edition 01/2024

5/6

Huba Control AG
Industriestrasse 175436 Würenlos, Switzerland
Tel. +41 56 436 82 00
info.ch@hubacontrol.com

Consultancy in your areahubacontrol.com/en/worldwide