PCB1 Key Operation Flowchart● About Setting Item

POWER ON

If no// (Data clear No) is selected: If
MODE is pressed, the unit returns to RUN mode.
If yes/ (Data clear Yes) is selected:
If MODE is pressed, the unit automatically
returns to RUN mode after data clear.

MODE

RUN Mode
Program control
Stop(in standby)

RUN(*)

STOP

Program control
RUN

Monitor Mode (Enabled during program control)
PTN

PTN

PTN

(1 sec)(*) Select a pattern with PTN, and press RUN.
Program control for the pattern will be performed.

Setting Mode

● Key Operation
Upper left: PV Display: Indicates setting characters.
Lower left: SV Display: Indicates factory default.
Right side: Indicates setting items.

Step 1 SV• Temp///0• A2// EV2 alarm value///0• Use ˄ and ˅ for settings, and register the settings with MODE or DISP.

Shaded setting items are optional, and appear only when the options are ordered.

• (*1) Available when 001 (High limit) to 012 (H/L limits with standby independent) are selected in [Event output
EV allocation].
• (*2) Available when 004 (H/L limits independent), 006 (H/L limit range independent) or 012 (H/L limits with standbyindependent) is selected in [Event output EV allocation].
• (*3) Available when 015 (Time signal output) is selected in [Event output EV allocation].
• (*4) Available when SV digital reception (Shinko protocol) is selected in [Communication protocol].
• (*5) Available when direct current or DC voltage input is selected in [Input type].
• (*6) Available when 001(High limit) to 012 (H/L limits with standby independent) – except [007 (Process high) and008 (Process low)] – are selected in [Event output EV allocation].

• MODE:

Moves to the next item, illustrated by an arrow.

• DISP:

Moves back to the previous item (Opposite to MODE).

• MODE (3 sec):

Press MODE for approx. 3 seconds.

• ˄ +MODE:

Press ˄ and MODE (in that order) together.

• ˄ + ˅ (3 sec):

Press ˄ and ˅ (in that order) together for 3 seconds.

• ˅ +MODE (3 sec):

Press ˅ and MODE (in that order) together for 3 seconds.

• ˄ + ˅ +MODE (3 sec): Press ˄ , ˅ and MODE (in that order) together for 3 seconds.
• ˄ + ˅ +STOP (3 sec): Press ˄ , ˅ and STOP (in that order) together for 3 seconds.
• RST:Returns to RUN mode from any setting items.

MODE

Pattern Setting Mode

Event Setting ModeControl Parameter Setting Mode

Wait Parameter Setting Mode

Engineering Setting Mode 1

Engineering Setting Mode 2
①

MODE

MODE

(*1)

MODE

MODE

(*)

MODE

⑥

MODE

MODE

(*1)

MODE

②

MODE

MODE

(*2)

MODE

MODE

MODE

MODE

(*3)

MODE

(*)

MODE

MODE
MODE

(*)

MODE

(*1)

MODE

③

MODE

MODE

④

MODE

MODE

(*3)

MODE

(*)

(*)
MODE
Setting items for
Step 2 to 10 appear.

MODE

Setting items for
Step 2 to 10 appear.
MODE

(*1)

MODE

(*)

MODE

MODE

(*)

MODE

MODE

MODE

MODE

MODE

MODE

MODE

(*5)

MODE

(*6)

MODE

MODE

(*1)

MODE

Returns to the 1st item.

MODE

MODE

(*6)

MODE

(*1)

MODE

(*) PTN/STEP Display indicatesthe selected pattern number.
PTN/STEP indicator lights up
the step number.

MODE

MODE

(*1)

MODE

(*1)

MODE

MODE

MODE

MODE

(*1)

MODE

MODE

MODE

MODE

(*1)

MODE

(*)

⑪(*3)

Returns to the 1st item.

Returns to the 1st item.

MODE
⑤

MODE

MODE

MODE

⑨

MODEMODE⑩MODE

PTN/STEP Display indicatesthe selected pattern number.
PTN/STEP indicator is unlit.

PTN/STEP Display indicates theselected pattern number.
PTN/STEP indicator lights up thestep number.

MODE

MODE

MODE

MODE

MODE

MODE

(*6)

MODE

MODE

(*1)

MODE

MODE

● Initial Setting
Before pattern setting and Event setting, perform initial setting as follows.
• Engineering setting mode 2: ① to ⑨
• Control parameter setting mode: ⑩，⑪

MODE

MODE

Loc5 Lock 5
Changeable in Set value lock
Sv// Step SV + Step timesvev

Step SV + Step time + EVvaluePTN/STEP Display andPTN/STEP indicator are unlit.

Returns to the 1st item.
MODE---- Pattern link Disabledchin Pattern link Enabled
AT Perform / Cancel---- AT Cancel
At// AT Perform
OUT2 cooling method
Air/ Air cooling
Oil/ Oil cooling
Wat/ Water cooling
Direct / Reverse action
heat Reverse control action
cool Direct control action
Step 1 to 10 Wait function Enabled / Disabled---- Disabled
Use/ Enabled
Set value lock---- Unlock
Loc1 Lock 1
Loc2 Lock 2
Loc3 Lock 3
Loc4 Lock 4

MODEReturns to the 1st item.
MODE

Pattern link

MODE⑧MODE

MODEMODE⑦(*) PTN/STEP Display indicates
the PID block number.
PTN/STEP indicator is unlit.

Stop bitalarm

Communication protocol
noml Shinko protocol
Svt/ SV digital transmission (Shinko protocol)
svtr SV digital reception (Shinko protocol)
moda Modbus ASCII mode
modr Modbus RTU mode
Communication speed//96 9600 bps/192 19200 bps/384 38400 bps
Data bit / Parity
8non 8 bits / No parity
7non 7 bits / No parity
8evn 8 bits / Even
7evn 7 bits / Even
8odd 8 bits / Odd
7odd 7 bits / Odd///1 1 bit///2 2 bits
Input typek//C K
k/ .C K
j//C J
r//C R
s//C Sb//C B
e//C E
T/ .C Tn//C N
Pl2C PL-IIc//C C(W/Re5-26)
Pt .C Pt100
jpt.C JPt100
Pt/C Pt100
jptC JPt100
k//f Kk/ .f Kj//f J-200 to 1370 °C-200.0 to 400.0 °C-200 to 1000 °C
0 to 1760 °C
0 to 1760 °C
0 to 1820 °C-200 to 800 °C-200.0 to 400.0 °C-200 to 1300 °C
0 to 1390 °C
0 to 2315 °C-200.0 to 850.0 °C-200.0 to 500.0 °C-200 to 850 °C-200 to 500 °C-328 to 2498 °F-328.0 to 752.0 °F-328 to 1832 °F

32 to 3200 °F
r//f R
32 to 3200 °F
s//f S
32 to 3308 °F
b//f B-328 to 1472 °F
e//f E-328.0 to 752.0 °F
t/ .f T
-328 to 2372 °F
n//f N
32 to 2534 °F
Pl2f PL-II
32 to 4199 °F
c//F C(W/Re5-26)-328.0 to 1562.0 °F
pt .f Pt100-328.0 to 932.0 °F
Jpt.f JPt100-328 to 1562 °F
Pt/f Pt100-328 to 932 °F
jptf JPt100-2000 to 10000
420a 4 - 20 mA-2000 to 10000020a 0 - 20 mA-2000 to 10000
0/1v 0 - 1 V-2000 to 10000
0/5v 0 - 5 V-2000 to 10000
1/5v 1 - 5 V-2000 to 10000
010v 0 - 10 V
Decimal point place///0 No decimal point//0.0 1 digit after decimal point/0.00 2 digits after decimal point

MODE(*4)

Returns to the 1st item.
PTN/STEP Display andPTN/STEP indicator are unlit.
0.000 3 digits after decimal point
Event output EV1 to EV3 allocation/000 No event/001 High limit alarm/002 Low limit alarm/003 H/L limits alarm/004 H/L limits independent alarm/005 H/L limit range alarm/006 H/L limit range independent alarm/007 Process high alarm/008 Process low alarm/009 High limit with standby alarm/010 Low limit with standby alarm/011 H/L limits with standby alarm/012 H/L limits with standby independent/013 Heater burnout alarm output/014 Loop break alarm output/015 Time signal output/016 Output during AT/017 Pattern end output/018 Output by communication command/019 RUN output＊/020 Heating/Cooling control output
Available only for Event output EV2 allocationEV1 to EV3 alarm value 0 Enabled / Disabled
No// Disabled
Yes/ Enabled
EV1 to EV3 alarm Energized / De-energizednoml Energized
revs De-energized
Event input DI1, DI2 allocation/000 No event/001 Pattern number selection/002 Direct / Reverse action/003 Program control RUN / STOP
/004 Program control Holding / Not holding/005 Program control Advance function
Transmission output type
Pv// PV transmission
Sv// SV transmission
Mv// MV transmission
Step time unit
Min/ Hours : Minutes
Sec/ Minutes : Seconds＊Power restore action
stop Stops after power is restoredcont Continues after power is restored
hold Suspends after power is restored
Program control start type
Pv// PV startPvr/ PVR start
Sv// SV start
Output status when input errors occur
Off/ Output OFFOn// Output ONError indication
No// Disabled
Yes/ Enabled