Instruction Manual

POWER SUPPLY for RD SERIES

RDP-24

No. RDP21JE3

2013.09

PrefaceThank you for purchasing the RDP-24, power supply for the RD series. Thismanual contains instructions for the mounting, functions, operations and notes foroperating the RDP-24. To ensure safe and correct use, thoroughly read andunderstand this manual before using this instrument. To prevent accidents arisingfrom the misuse of this instrument, please ensure the operator receives thismanual.

Caution• This instrument should be used in accordance with the specifications described inthe manual. If it is not used according to the specifications, it may malfunction orcause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed,serious injury or accidents may occur.
• The contents of this instruction manual are subject to change without notice.
• Care has been taken to assure that the contents of this instruction manual arecorrect, but if there are any doubts, mistakes or questions, please inform oursales department.
• This instrument is designed to be installed on a DIN rail within a control panel.
Measures must be taken to ensure that power terminals or other high voltage
sections cannot be touched.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos Co., Ltd. is not liable for any damage or secondary damage(s)incurred as a result of using this product, including any indirect damage.

Wiring Precautions

Caution• Do not leave wire remnants in the instrument, because they could cause a fire
and/or a malfunction.
• When wiring terminals, use ferrules with an insulation sleeve and crimping
pliers made by Phoenix Contact GMBH & CO.
• This instrument has no built-in power switch, circuit breaker or fuse.
It is necessary to install them near the instrument.
(Recommended fuse: Time-lag fuse, rated voltage 250V AC, rated current 2A)• For the AC power source, connect to the terminals specified in this manual.
If the AC power source is connected to any other terminals, this unit will burnout.
• Keep the output wires and power line separate.

Operation and Maintenance Precautions

Caution• Do not touch live terminals. This may cause electric shock or problems in operation.
• Turn the power supply to the instrument OFF before cleaning.
Working on or touching the terminal with the power switched ON may result insevere injury or death due to electric shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)

1. Name and Functions of Sections

Safety Precautions (Be sure to read this before using units)PWR indicator (Green): Power indicator
Lights when power is turned ON.

The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on circumstances, procedures indicated by
Caution may cause
serious results, so be sure to follow the directions for usage.
which may lead to dangerous conditions andWarning Procedurescause death or serious injury, if not carried out properly.
which may lead to dangerous conditions andCaution Procedurescause superficial to medium injury or physical damage or may
degrade or damage the product, if not carried out properly.

Warning

Spring type plug

• To prevent an electric shock or fire, only Shinko or qualified service personnel
may handle the inner assembly.
• To prevent an electric shock, fire or damage to the instrument, parts
replacement may only be undertaken by Shinko or other qualified service
personnel.

2. Mounting
2.1 External Dimensions (Scale: mm)
DIN rail75

Safety Precautions• To ensure safe and correct use, thoroughly read and understand this manual
before using this instrument.
• This instrument is intended to be used for industrial machinery, machine
tools and measuring equipment. Verify correct usage after purpose-of-use
consultation with our agency or main office. (Never use this instrument for
medical purposes with which human lives are involved.)• External protection devices such as protective equipment against excessive
temperature rise, etc. must be installed, as malfunction of this product could
result in serious damage to the system or injury to personnel. Also proper
periodic maintenance is required.
• This instrument must be used under the conditions and environment
described in this manual. Shinko Technos Co., Ltd. does not accept liability
for any injury, loss of life or damage occurring due to the instrument being
used under conditions not otherwise stated in this manual.

(Fig. 1-1)

CautionThis instrument is intended to be used under the following environmental
conditions (IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable or explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of -5 to 55(23 to 131 ) that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85%RH• No large capacity electromagnetic switches or cables through which large
current is flowing• No water, oil or chemicals or where the vapors of these substances cancome into direct contact with the unit
• Take note that the ambient temperature of this unit must not exceed 55(131 ) if mounted within the control panel. Otherwise the life of electroniccomponents (especially the electrolytic capacitors) may be shortened.
Note: Avoid setting this instrument directly on or near flammable material eventhough the case of this instrument is made of flame-resistant resin.

85

18.33

4(Fig. 2.1-1)
2.2 Mounting to the DIN RailNotice• Mount the DIN rail laterally.
• The right shows a DIN rail

corresponding to the RDP-24.

• Use end plates at both endsof the unit for fixing.

35 0.1mm
23mm or more

7.5mm or more

Installation Precautions

17.5

Mounting screw height
3.4mm or less

To avoid this instrument from being used as a component in, or as being utilized
in the manufacture of weapons of mass destruction (i.e. military applications,military equipment, etc.), please investigate the end users and the final use ofthis instrument. In the case of resale, ensure that this instrument is not illegallyexported.

4

Caution with Respect to Export Trade Control Ordinance(Fig. 2.2-1)
Recommended Fastening Plate
Manufacturer
ModelOmron corporation
End platePFP-M
IDEC corporation
Fastening plate BNL6
Panasonic Electric Works Co., Ltd.
Fastening plate ATA4806• Do not confuse the top and bottom of this unit.
• When mounting to or removing the unit from the DIN rail, the units must be
slanted a little. Therefore do not mount any other instruments within aspace of 50mm either on the upside or the downside of the unit, in
consideration of wiring space and heat radiation.

(1) Pull the lock lever of the unit down ( 1 ), hook the unit to the DIN rail at anangle of approx. 20 degrees upward ( 2 ), and mount it ( 3 ).

3.4 Wiring Example
For the wiring, refer to the Instruction Manual for each instrument.
RD-500, RD-700
100 to 240V AC

Indicator/Controller(ACS-13A- /A)

RDP-247

Power terminals

65+
Input terminals

RD-500orRD-700(Fig. 2.2-2)

4-3[PWR-]-[PWR+]+(2) Push the lock lever of the unit up until a clicking sound is heard ( 4 ).

21(Fig. 3.4-1)
RD-600100 to 240V AC●●RDP-24

Indicator [JIR-301-M (0 to 10V DC input)]/Controller

76Power terminals

54(Fig. 2.2-3)

Input terminals

32(3) Fix both ends of the unit with the fastening plates (Refer to the recommendedfastening plates.).

3. Wiring

1-++8..36VDC

3.1 Ferrules
When using ferrules, use the following recommended ferrules and crimping
pliers made by Phoenix Contact GMBH & CO. (Table 3.1-1)(Table 3.1-1)
Ferrules withConductor crossCrimping pliers
insulation sleevesections
AI 0.25-6 BU
0.2 to 0.25mm2
AI 0.34-8 TQ
0.25 to 0.34mm2
AI 0.5-8 WH
0.34 to 0.5mm2
CRIMPFOX ZA 3
AI 0.75-8 GY
0.5 to 0.75mm2
CRIMPFOX UD 6
AI 1.0-8 RD
0.75 to 1.0mm2
AI 1.5-8 BK
1.0 to 1.5mm2
AI 2.5-8 BU
1.5 to 2.5mm2765

OUTPUT 2+
OUTPUT 1+GND

○ ○

GND

OUT-mV/mA○ ○ ○ ○

RD-600 (Electronics Module)(Fig. 3.4-2)

4. Specifications
RatingSupply VoltageAllowable Voltage Range
Output Voltage, Current

100V to 240V AC 50/60Hz
85 to 264V AC
Output 1: 24V 30mA DC, Output 2: 18V 30mA DCPerformance
Output Voltage Accuracy

Output 1: 5% or less,Output 2: 15% or less (No load: +50% or less)
Supply Voltage Fluctuation
1% or less at 100V to 240V AC
Load Fluctuation
Output 1: 5% or less, Output 2: 10% or less
Ripple Noise
300mVp-p or less
Overcurrent Protection
Automatic return
Overvoltage Protection
None

3.2 Terminal Arrangement
POWER SUPPLY
100 to 240V AC+4

Insulation, Dielectric Strength
Insulation Resistance
Output 1 – Output 2 – Power: 10M or more, at500V DC
Dielectric Strength
Output 1 – Output 2 – Power: 2.0kV AC for 1 min.

321(Fig. 3.2-1)

3.3 Wiring of Spring Type PlugFor spring type plugs, insert the wire while pressing the lever (orange color)
with a flat blade screwdriver. (Fig. 3.3-1)

Other
Mounting
Applicable Products
Power Consumption
WeightDIN-rail (Within a control panel or equivalent)
RD-500, RD-600, RD-700 (For specified use only)
Approx. 7VA
Approx. 80g

Inquiries
For any inquiry about this unit, please contact the vendor where you purchased
the unit or our agency after checking the following.
(e.g.)• Model ---------------------- RDP-24• Serial number ----------- No.123456789
In addition to the above, please let us know the details of malfunction, if any, andthe operating conditions.

The wire to which theferrule with an insulation
sleeve is clamped

Insert the wire whilepressing the lever(orange color) with aflat blade screwdriver.
(Fig. 3.3-1)

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, Japan
http://www.shinko-technos.co.jp
URL:
Tel : +81-72-727-6100
E-mail:overseas@shinko-technos.co.jp
Fax: +81-72-727-7006