USB Digital Storage Oscilloscope
Seriesathtepocsolli
plications
paetcaerscootuloyatsiallow
Dig★ Provides a digital oscilloscope function to your computer★ Allows you to create your own applications with the Software

Development Kit (SDK) provided for free
★ Equipped with the FFT function★ Also, we can quickly handle custom speciﬁcations
PA-S2000

Application software: PandaScope

Carrying bag (included)Standard price (including tax)
PA-S2000:Digital Storage Oscilloscope(equipped with 100 kB memory)PA-S2000/E :PA-S2000 equipped with 1 MB memory

92,400 yen
108,150 yen[Options]
PA-S2000/PRB: x1/x10 probe (2 in 1 set)
PA-S2000/P100: x100 probe (2 in 1 set)

PA-S2000
PA-S2000/E (dual cable type)

30,450 yen
37,800 yenSmall and Lightweight Digital Oscilloscope
SeriesPA-S2000 provides a digital oscilloscope function to your computer via USB connection. It is small, lightweight and very portable, and PC functions suchas waveform storage and analysis can be used. We provide an environment in which you can create your own application, by providing a windows API.

Features- Small and lightweight- Provides an analog frequency bandwidth of 200 MHz and sampling rate of
100 MS/sec (2 channels at the same time). And, 10 GS/s for equivalentsampling.
- Because the power is supplied from the USB port, external power supply,
such as AC/DC adapter, is not necessary.
- Fast data transfer to your computer can be accomplished through the USB
port, providing a comfortable screen refresh rate.

- The unit is equipped with a memory that can store 100 k points (1 Mpoints for PA-S2000/E) of data. By using with plenty of trigger functions,single-shot events and complicated waveforms can be easily captured
and observed.
- With the attached Windows application, complicated functions can beeasily operated.
- If desired, we will provide the SDK for Windows free of charge (APIspeciﬁcations and sample source code in VB, VC, Delphi, etc.)

Speciﬁcations
HardwareVertical axis

Environmental characteristics

Frequency band
Sampling rate

DC coupling: DC to 200 MHz (-3 dB ± 1 dB)
AC coupling: 3.5 Hz to 200 MHz (-3 dB ± 1 dB)
Real sampling: 100 M samples/sec max (2 channels at the same time)
Equivalent sampling: 10 G samples/sec max (2 channels at the same time)

Operating temperature range

0°
C to 60°
C

Speciﬁcation guaranteed operating
temperature range
Speciﬁcation guaranteed operating
humidity range

10°
C to 35°C
85% or less (non condensing)

Signals to be measured Limited to the low voltage circuit on the secondary side

Number of input channels

Unbalanced, 2 channels

Interface

Vertical axis display range

8 div (data content is 10 div)

Interface speciﬁcations

Compliant with USB2.0 full-speed (12 Mbps) speciﬁcation

Vertical axis sensitivity 10 mV/div to 2 V/div, in steps of 1 to 2.5(scale range)(Scale is 10 times of this for 10:1 probe, and 100 times of this for 100:1 probe)

Consumption current

500 mA or lessOthers

GND of USB and measuring terminal is not insulated

Vertical axis resolution 1024 points (10 bits) or 512 points (9 bits)

Dimensions and weight

Vertical axis amplitude accuracy (DC) ±3% (central value)

Dimensions

145 mm (L) x 100 mm (W) x 23 mm (H), not including BNC terminal

Input coupling

DC, AC, GND (software)

Weight (main unit)

Approx. 280 g

Input impedance

1 MΩ ± 3%, approx. 20 pF

Accessories

Maximum input voltage

50V (DC + AC peak)

Position variable range±5 div- USB cable (1.5 m)- Written guarantee- Support CD- Carrying bag

600 mA or less

- Dual USB cable (70 cm)- Written guarantee- Support CD- Carrying bag

Trigger
Trigger signal source

Signal from CH1, CH2 or external trigger input

External trigger input impedance 1 MΩ ± 3%, approx. 20 pF

SoftwareExternal trigger amplitude ±2.5 Vp-pTrigger voltage setting range±5 div (ﬁxed to ±2.5 V for external trigger input)

Trigger slope polarity

Positive-going edge, negative-going edge

Trigger coupling

DC, HFrej (high-frequency rejection), LFrej (low-frequency rejection)

Digital partTime axis range

Memory size 10 k points: 1 ns/div to 10 s/div

SoftwareMemory size 10 k points: 1 ns/div to 10 s/div

Display waveform memory

Select from 10 k points or 100 k points Select from 10 k points, 100 k points or 1 M points

Cursor measurement

ΔV, ΔT, Δ1/T

File handlingWaveform data: stored in storage in CSV format

Calculation function

FFT, X-Y, addition, subtraction, multiplication and division

Waveform display image: stored in BMP/JPG format

Memory size 100 k points: 1 ns/div to 100 s/div Memory size 100 k points: 1 ns/div to 100 s/div

Operating environment

In steps of 1 to 2.5

OSMemory size 1 M points: 1 ns/div to 1000 s/div
In steps of 1 to 2.5

Sampling mode

Single shot (stops after capturing 1 point of memory),roll (continuous)

Memory length

100 k (100,000) points/chTrigger position

Anywhere within the 100 k points Anywhere within the 1 M points

Acquisition mode

Sample, average, envelope

1 M (1,000,000) points/ch

Windows 2000/XP/Vista/7

Display size

1024 x768 or higher

CPU (recommended)

Pentium 1.5 GHz or higher

Memory (recommended)

768 MB or more

1.5 GB or more

USB port

USB 1.1 or USB 2.0

USB 1.1 or USB 2.0
Two USB ports are used

Calibration signal
Signal waveform

Square wave

PA-S2000 Options

Voltage

1 Vp-p ± 2% Positive polarity

Frequency

1 kHz ± 0.1%

- PA-S2000/PRB: 250 MHz passive probe (with x1/x10 change-over switch),
1.2 m, 2 in 1 set- PA-S2000/P100: 250 MHz passive probe (x100), 3 m, 2 in 1 set● A demo unit is available free of charge. For more information, please contact your dealer or our company.
● Company names and product names are registered trademarks of their respective holders. The above contents
are subject to change without notice for improvement of products.

Ogasawara Bldg. 2F, 16-13-1, Nagai, Morioka-shi, Iwate 020-0834 Japan
Phone: +8119-637-8330 / Fax: +8119-637-8331

If you have any question, please contact our company throughthe contact in the website or by fax.