SFLC-212-173

INSTRUCTION MANUALPOWER LINE MULTI-METER
SFLC-110L
1φ2W / 1φ3W / 3φ3W
ANALOG OUTPUT
HARDWARE MODEL FSFLC-212-173

Thank you for your purchase of our product.
Read this instruction manual carefully before installation, wiring, and using this product.

Safety concerns
 Environment and conditions of usage
Please be sure to use this product in a place that meets the following conditions.
In places that do not meet this condition, it may cause malfunction or failure and product life decline.
 Within the range of ambient temperature -10 to +55 ℃, humidity 85％ RH or more.
 Place free of dust, corrosive gas, salt and oily smoke. (Corrosive gas：SO2 / H2S, etc.)
 Location that is not affected by vibration and shock.
 Location that is not affected by external noise.
 Altitude 2000m or less.
 If this unit directly measures an inverter output of cycle control, SCR phase angle control or PWM control,
an error may increase due to its operation principle.
 The precautions at the case of using by outdoor panel.
 These products are not a dustproof construction, waterproof construction, and splash proof construction.
Please avoid the place with much dust. Moreover, please install in the place which requires neither rain
nor waterdrop.
 Please do not install in the place where sunlight hits directly.
Discoloration and degradation of a name plate, and cover is deformed by the surface temperature rise.
 Installation and wiring
Installation and wiring, refer to the instruction manual, please be conducted by engineers.
And, please observe the following notes.
 Please wire after the connection diagram is checked.
 Please forbid a hot line work.
 Please use the size of the electric wire that is suitable for conducting current.
 Please securely tighten the terminal screws. Please make sure that forgotten tighteningof the terminal screw is not.
 Preparation before use
At the case of connect this product to the main power supply directly, please put the suitable fuse to the outside.
This product must be set before use. Reading this instruction manual, please set correctly.
 About dew condensation
If the temperature and humidity of an installation change rapidly when a product is a non-energization, thewaterdrop by dew condensation may adhere to a display inner side. (The display filter and the LCD surface
stick and the pattern of the shape of a circle or an ellipse occur.)This phenomenon is not trouble. It will disappear, if a control power supply is applied and about 2 hour passes.
 Maintenance Inspection of energized is dangerous.
 There is no parts to be replaced on a regular inspection.
 Please wipe off lightly with the dry soft cloth. When it wipes with the damp cloth or the dry cloth strongly,
a surface is damaged. And, the character of a name plate may disappear.
Please do not use the organic solvent, chemicals, cleaners, etc., such as an alcohol, for cleaning.
The liquid crystal display (LCD) may light during cleaning on the LCD face. However, this phenomenon is causedby the static electricity that may be produced in the filter, and it does not show any trouble.
Leave the unit as it is for a while, and the display goes out due to natural discharge.
Do not press the LCD face strongly, otherwise it may be broken. When the filter has been pressed, it may touch
the LCD face to stain the LCD face. However, this phenomenon does not show any trouble, but it is caused bya change of the ambient environment or the like. The LCD face may be restored to its original condition after
a while during the use as it is.

1

SFLC-212-173

 Storage
Please be storage in a place that meets the following conditions.
 Within the range of ambient temperature -20 to +70 ℃.
 The average temperature (day) 40 ℃ or less.
 Place free of dust, corrosive gas, salt and oily smoke.
 Location that is not affected by vibration and shock.
 The aluminum electrolytic capacitor is used for a product. Please do the energization of the power
supply within one year after shipment.
 Countermeasures against troubles
If the product has failed, we can repair pick up the product.
 Proper disposal
Please dispose of this product as industrial waste (noncombustible).
Mercury parts and a nickel-cadmium battery are not used for this product.
 Warranty period
The warranty period of this product is for one year after supplying the appointed place.
 Operation
Be careful with the following cautions during use.
 Use the input within the rated range. Be careful since negligence of this caution may cause troubles of
the unit.
 There is a function to hold the maximum value and the minimum value with a measurement factor in this product.
A blackout is guaranteed and this value isn't also cleared by a power supply reset. However, the minimum value
may be updated in case input is not applied to a power up. For this reason, in order to make the past minimum
value hold by powering on, please apply input within 1 second after switching on a power supply.
 The maximum value, a minimum value measurement factor
Maximum value
Minimum value
Measurement factor
measurement
measurement
Voltage，Current ，Demand current ，Active power ，Demand power ，○○Reactive power ，Power factor ，Frequency
 Be careful not to touch any terminal when power is applied to the unit.
 Don’t disassemble or modify this unit without any previous permission of our company,
otherwise the warranty does not apply to the unit any more. Also, modifications may cause
troubles, a fire, or other accidents. For specifications change, etc., please contact us.
Please forbid a hot line work.
 SettingThis unit requires setting and confirmation of the measuring range, etc. before use.
Wrong setting, if any, causes malfunction of the unit. If setting should be wrong, neither measurement nor output
becomes correct. Carefully read the instruction manual before setting the unit.
 Default setting
The default settings are as specified below at the delivery time. Set them according to the working conditions.
The input circuit of this product is the common use of 3-phase 3-wire (3φ3W), single-phase 2-wire (1φ2W), andsingle-phase 3-wire (1φ3W). In case an input circuit is designated at the case of an order, it is shipped by thedefault value of the designated input circuit. And, in case it does not do designation of the input circuit (withno designation), it is shipped by the default value of 3-phase 3-wire (110V).
The unit will be delivered with your specified setting values, if so specified.

2

SFLC-212-173

3-phase 3-wire№

1

Display
combination

2

Alarm output(１ )3

Demanddetection

4

Instant
measurement
detection

5

Backlight

6

Pattern
Main monitor
Sub monitor (Left)
Sub monitor (Center)
Sub monitor (Right)
Bar graph
FactorReset form
Contact delay time
Upper limit <H>
Demandcurrent Interval
Upper limit <H>
Interval
Demandpower
Operation form

Pattern 1
A(V)
V(UV)
W
Wh
A(V)
DA
AUTO
0 second
80.0A
0 secondOFF
0 second
Operating system
according with bimetallic
type.

Single-phase3-wire(U-N-W)
Pattern 1
A(U)
V(UN)
W
Wh
A(U)
DA
AUTO
0 second
400.0A0 secondOFF
0 second
Operating system
according withbimetallic type.

Voltage upper limit

OFF

OFF

OFF

Voltage lower limit

OFF

OFF

OFF

ActionBrightness (４)Voltage range
Digit number of voltage range
Current range
Current display intrinsic
sensitivity
Digit number of current range

AUTO
3（Middle）
6600V
4 digits
100.0A

AUTO
3（Middle）
110.0V
4 digits
500.0A

AUTO
3（Middle）
3300V
4 digits
50.00A

100.0A

500.0A

50.00A

4 digits

4 digits
One-way
deflection

4 digits

Setting item

Measurement
range

8
9

220V input

220.0V
4 digits

110V input

220V input

Pattern 1
A
V
W
Wh
A
DA
AUTO
0 second
40.00A0 secondOFF
0 second
Operating system
according with bimetallic
type.

220.0V
4 digits

Active power polarity

One-way deflection

Active power range

1200kW( /1kW)

40.00kW( /2kW)

100.0kW ( /1kW)

150.0kW( /500W)

10.00kW( /1kW)

Digit number of active powerrange

4 digits

4 digits

4 digits

4 digits

4 digits

Reactive power range

LEAD 600.0
to 0.0 to
LAG 600.0kvar( /500var)

LEAD 20.00
LEAD 50.00 to 0.00
to 0.00 to
to LAG 50.00kvar
LAG 20.00kvar( /500var)( /1kvar)

LEAD 75.00
to 0.00 to
LAG 75.00kvar( /250var)

LEAD 5000
to 0 toLAG 5000var( /500var)

Digit number of reactive
power range

4 digits

4 digits

4 digits

4 digits

4 digits

LEAD0.500 to
1.000 to LAG0.500
45.0 to 65.0Hz
3 digits
A(U)
V(UN)
W
100.0％
100.0％
50.0％
OFF
Wh

LEAD0.500 to 1.000
to LAG0.500
45.0 to 65.0Hz
3 digits
A
V
W
100.0％
100.0％
50.0％
OFF
Wh

Power-factor range

7110V input

Single-phase

Frequency range
Digit number of frequency range
Output factor 1
Output factor 2
Output factor 3
AnalogoutputCurrent
Output(１ )intrinsic
Active power
sensitivity Reactive power
Low input cut
Pulse output Factor(１ )Pulse unit
External operation input (１)LEAD0.500 to 1.000 to
LAG0.500
45.0 to 65.0Hz
3 digits
A(V)
V(UV)
W
100.0％
100.0％
50.0％
OFFWh
10kWh/pulse

Alarm reset

3

0.1kWh/pulse

One-way deflection

1kWh/pulse

1kWh/pulse

Alarm reset

Alarm reset

0.1kWh/pulse

SFLC-212-173

ON
ON
ON
ON
ON
ON
ON
ON
ON

Single-phase3-wire(U-N-W)
ON
ON
ON
ON
ON
ON
ON
ON
ON

ON
ON
ON
ON
ON
ON
ON
ON
ON

ON

ON

ON

ON

ON

ON

ON

ON

ON

1φ3W (U-N-W)
150V
Generalmeasurement
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

1φ2W
110V

3-phase 3-wire№

Setting item

11

Input
circuit

Voltage
Current
Demand current
Active power
Demand power
Reactive power
Power-factor
Frequency
Watt-hour of power receiving
Watt-hour of powertransmission
var-hour of power receiving
var-hour of powertransmission
Phase line change (６)Input voltage (７)12Measurement

Tidal current measurement

13

Analogoutputadjustment
(５ )

10

Measurement
display
ON/OFF

Output 1
Output 2
Output 3

Bias adjustment
Span adjustment
Bias adjustment
Span adjustment
Bias adjustment
Span adjustment

110V input

3φ3W
110V

220V input

220V

General measurement
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

Single-phase
110V input

220V input

220V

General measurement
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

Note(１) A setting item is not displayed in case there is no corresponding option.
Note(２) When the input circuit phase line switching setting is changed, the setting values of setting 1 andsetting 2 (No. 1 to 10 in the table) return to the default setting values of the switched phase line.
However, the value of setting 3 (No. 11 to 13 in the table) does not return to the initial setting.
Note(３) When phase line change setting of an input circuit is set as 3φ3W (or 1φ2W) and the input voltage
setting is changed, the voltage range returns to the default value of the phase line.
(For example：In case of 3φ3W, 6600V at the case of 110V setting, 220.0V at the case of 220V setting.）

4

SFLC-212-173

Content

Safety concerns ･･････････････････････････････････････････････････････････････

1

1. Product outline ･･･････････････････････････････････････････････････････････
1.1 Usage of product ･･･････････････････････････････････････････････････････
1.2 Features of product ････････････････････････････････････････････････････
1.3 Composition of type ････････････････････････････････････････････････････

6
6
6
6

2. The name and function of each part ････････････････････････････････････････

7

3. Preparation
3.1 Installation ･･･････････････････････････････････････････････････････････
3.2 Connections ････････････････････････････････････････････････････････････

8
9

4. Operation ･････････････････････････････････････････････････････････････････ 10
4.1 The screen change and function by switch operation ･････････････････････ 11
4.2 The kind of display ････････････････････････････････････････････････････ 12
4.2.1 Measurement display ･････････････････････････････････････････････････ 12
4.2.2 Alarm detection display ･････････････････････････････････････････････ 13
4.2.3 Setting display ･････････････････････････････････････････････････････ 15
4.3 Operation ･･････････････････････････････････････････････････････････････ 16
4.3.1 The main monitor display-element change ･････････････････････････････ 16
4.3.2 Phase (line) display change ･････････････････････････････････････････ 16
4.3.3 Enlarged display of integral power consumption ･･････････････････････ 17
4.3.4 Setting value check ･････････････････････････････････････････････････ 17
4.3.5 Setting mode ････････････････････････････････････････････････････････ 18
4.3.6 Reset ･･･････････････････････････････････････････････････････････････ 19
5. Setting
5.1 Function table ･････････････････････････････････････････････････････････ 22
5.2 Setting table ･･････････････････････････････････････････････････････････ 26
5.3 Setting in detail explanation ･･････････････････････････････････････････ 33
5.3.1 Setting mode 1 ･･････････････････････････････････････････････････････ 33
5.3.2 Setting mode 2 ･･････････････････････････････････････････････････････ 41
5.3.3 Setting mode 3 ･･････････････････････････････････････････････････････ 49
6. Specification
6.1 Specification and intrinsic error ･･････････････････････････････････････ 53
6.2 Performance ････････････････････････････････････････････････････････････ 55
6.3 Option ･････････････････････････････････････････････････････････････････ 57
7. Maintenance and check7.1 Trouble shooting ･･･････････････････････････････････････････････････････ 58
7.2 Test ･･･････････････････････････････････････････････････････････････････ 59
Appendix table 1 to 3

5

SFLC-212-173

1. Product outline
1.1 Usage of productThis single unit can measure and monitor voltage ×3, current ×3, demand-current ×3, active-power, demand
active power, reactive-power, power-factor, frequency, watt-hour, var-hour.
The measurement monitor of an initial power receiving circuit, an energy conservation power monitor, a demandcurrent measurement monitor, etc. are adapted for various usages from a low tension circuit to a high tension
circuit.
The intensive monitor doubled with the system by option (analog output, alarm output, pulse output,
external operation input) addendum is possible.
1.2 Features of product High-performance products which collected the function of the power line multi-meter (QLC-110L, DLC-110L)
of our company.
 This product is 3-phase 3-wire, single-phase 2-wire, single-phase 3-wire common type.
Product inventory can be standardized.
 Bar graph 1 measurement and digital 4 measurement are displayed simultaneously.
 Analog output 3 circuit and a pulse output and a contact-output can be taken out. (Option)
And, An output factor can be selected by setting.
 External operation inputs are possible of reset. (Option) Choice of an alarm output, the maximum/minimum value,
and an alarm output, and the maximum/minimum value is possible at setting.
 Power supply is AC85 to 264V，DC80 to 143V (for both AC and DC uses).
 The mounting method of this unit is compatible with the mounting method of conventional 110 squaremechanical meter. This unit is mounted at 2 diagonal points.
 Integrated value of Wh and varh can indicate expansion to 3rd digit below the decimal point.
 A tidal current measurement (output 2 quadrant) change is possible for var and cosφ.
 With backlight (white LED backlight) function.
In addition, the selection of the on / off / auto off and, you can set the brightness.
 LCD can be chosen from 2 kinds, the type to see from the top and the type to see from the bottom.
(Please designate it at an order.)1.3 Composition of type
TYPE

Specification code

SFLC-110L- □□□□□-□□□
Installation position
１：Upper installation model
２：Lower installation model
Ｆ：Wide viewing angle model
OptionSpecification
code
０－００

External operation
input
Nothing

Analog output orcommunication output
Nothing

１－１１
１－Ｍ１
１－Ａ１

One circuit
One circuit
One circuit

DC4 to 20mA×3 circuits
Modbus RTU (RS-485)
Protocol A (RS-485)

Auxiliary power supply
１：AC85 to 264V, DC80 to 143V for both AC and DC uses
Input range
Ｆ：Common use for AC150V, AC300V.

AC5A

Input circuit
Ｆ：Common use for 1φ2W, 1φ3W, 3φ3W.
Hardware model
Ｄ：R-S-T，White backlight
Ｅ：R-Y-B，White backlight
Ｆ：U-V-W，White backlight6

Contact output
Nothing
Pulse ＋ Alarm
Pulse ＋ Alarm
Pulse ＋ Alarm

SFLC-212-173

2. The name and function of each partBar graph display
The measurement value of the main monitor
is indicated by the analog.
(Setting which does bar graph displayof the measurement value of sub monitor
is also possible.)

Scale markings
Set the automatically by measurement-range
setting.
Upper limit (or lower limit) setting index
An upper limit (or lower limit) set point isdisplayed.

Digital display
Measurement monitor can watch
4 elements at the same time.

Unit display
Set the automatically by measurement-range
setting.

･Main monitor･Sub monitor (Right)･Sub monitor (Center)･Sub monitor (Left)

Multiplying factor display
Displays on the lower right of the main monitor
at the time of watt-hour and var-hour display.

SETDISPLAY

The switch from which integrated
value of the amount of electric
power is switched to normal display(5 digits of integer) and expansion
indication (integer 2 digits ＋ below
decimal point, 3 digits) variously.
If it is not operated for 10 minutes after adisplay change, it will usually return to adisplay. It is used also as a switch whichchanges to setting mode. If it continues
pushing 3 seconds or more, it will change tosetting mode.
In setting mode, it is used as a switch that
determines a set point.

The switch that changes a phase (line)
display of current (voltage).
If it is not operated for 10 minutes after adisplay change, it returns to the original
phase (line) display. In setting mode, it isused as a switch that terminates setting mode.

－＋

The switch to which measurement displays
element of main monitor is changed.
If it is not operated for 10 minutes after a display
change, it returns to the original measurement display
factor. In set mode, it is used as a switch that changes
a setting value.

RESET/SHIFT

MAX/MIN

If this switch is pushed 1 second or more
continuously, it will reset an alarm, the maximum
value, and the minimum value (factor which it isdisplaying on the main monitor).
In setting mode, it uses it as a switch to which it
moves a setting item (move up).

The switch to which general measurement display (usually)
and the maximum minimum measurement display are changed.
In setting mode, it uses it as a switch to which it moves asetting item (move down).
MODE

In setting mode, it is used as a switch that changes a setting
item.

7

SFLC-212-173

3. Preparation
3.1 Installation
Mount the unit by the attached M5 nuts to a panel of thinner than 10mm, referring to the following external
dimensions drawing and panel cutout. Fasten these nuts with tightening torque 2.0 to 2.5N･m.
● Dimension diagram● Caution on handling
Mount the LCD to obtain an optimum angle, since the contrast changes according to the monitoring angle.
(1) For upper installation model(2) For lower installation model(3) For wide viewing angle model
75°

60°
10°
10°
60°

75°
60°

(Side view)(Top view)

60°

60°

(Side view)(Top view)● Installation(1) A product is put in a cut hole of a panel from a front.
A body is inserted until it exceeds retaining stopperof the lower base.
(2) Please fix a product certainly with attached M5flange nut for installation. Please give atightening torque as 2.0 to 2.5N･m.

8

75°

60°

(Side view)(Top view)

75°

SFLC-212-173

3.2 Connections
Please perform connection after referring to the following wiring diagram.
● Connection drawing (６)(1) 1φ2W，Analog output.
UV(2) 1φ3W, Analog output.
UNW

VT

CT

CT
LOAD

AUX.SUPPLY

AUX.SUPPLYLOADANALOG OUTPUT 1
ANALOG OUTPUT 2
ANALOG OUTPUT 3

ANALOG OUTPUT 1
ANALOG OUTPUT 2
ANALOG OUTPUT 3

PULSE OUTPUT
ALARM OUTPUT

PULSE OUTPUT
ALARM OUTPUT
EXT.INPUT

EXT.INPUT(3) 3φ3W, Analog output
UVW

VT

CT

LOAD

Note(４) Analog output, pulse output, alarm
output, external operation input is an
option.
Note(５) By setting, it can switch an external
operation input.
Alarm reset. Maximum, minimum value reset.
Alarm and maximum, minimum value all
reset.
Note(６) In case of low-voltage circuit, secondaryside earthing of VT and CT is unnecessary.
And, VT is unnecessary in case it used 110V
or direct 220V.

AUX.SUPPLYANALOG OUTPUT 1
ANALOG OUTPUT 2
ANALOG OUTPUT 3
PULSE OUTPUT
ALARM OUTPUT

EXT.INPUT● Caution on connection(1) Mount the terminal cover without fail for safety after the end of connections.
(2) Separate the input wiring and output wiring from each other without fail, and take a preventive measureagainst malfunction due to external noises.
(3) Connect the grounding terminal E (No. 7 terminal) to the ground without fail for enhancing the shield effect.
Keep the grounding resistance between the grounding terminal and the ground to be lower than 100Ω.
(4) Keep a distance of more than 30cm between this unit and the circuit breaker as well as between this unit
and the relay contact signal line.
(5) No protection is necessary for this unit even if the transmission line may be affected by an induced lightning
surge or the like when transmitting an analog output to the receiver. Mount an about 500V arrester orthe like between the line surge protector and the ground as well as between the transmission line andthe ground on the receiver side for the purpose of protecting the devices on the receiver side.
(6) It is recommended to mount a surge killer outside when connecting an inductive load to the pulse output
and alarm output. If no surge killer is mounted, the contact life may shorten.
9

SFLC-212-173

4. Operation● The function of switch

SwitchFunction

SET

The integrated value of electric energy is changed to the usual display and an enlarged display.
If it continues pushing 3 seconds or more, it will change to setting mode.
In setting mode, it is used for the determination of a set point.

RESET/SHIFT

Various kinds of alarms are reset. The maximum value and the minimum value are reset in the maximum
minimum measurement display. In setting mode, it uses it as a switch to which it moves (move up)
a setting item.
In setting mode, it is used for the change of a setting item.

MODE
MAX/MIN

The usual measurement display and maximum value or minimum value display are changed.
In setting mode, it uses it as a switch to which it moves (move down) a setting item.

＋ ，－

The measurement display element of the main monitor is changed.
In setting mode, it is used for change of a set point.

DISPLAY

A phase (between lines) display of current (voltage) is changed. It is used in case it terminatessetting mode. And, it is used in case it returns the display combination of a measurement factor.

● Convenient functions(1) In case a measurement change or a phase change is performed and the original screen composition is notclear anymore, DISPLAY is pushed for more than 3 seconds or it's no-operation for 10 minutes and returns
to original screen structure.
(2) If a measurement change is performed and the original main monitor display is not clear anymore.
It continues pushing a ＋ or － more than 3 seconds, or returns to the original measurement display
factor by no operation for 10 minutes.
(3) Even if it stops operation with setting mode, it returns to the display mode in 10 minutes.
● 7 segment displays
This product shows the guidance in various setting using 7 segment displays besides a display of a measurement
value. A digital readout and 7 segment displays corresponding to each alphabet are shown in the following.

A

B(b)

C

D(d)

E

F

G

H

I

J

K

L

M

Y(y)

Z

Non-dis Non-dis
play
play

N(n)

O(o)

P

Q(q)

R(r)

S

T(t)

U(u)

V

W

X
Non-dis
play

0

1

2

3

4

5

6

10

7

8

9

SFLC-212-173

4.1 The screen change and function by switch operation.
This product changes various screens by switch operation.
operation is explained.

Here, the change step of the screen by switch
Wh,varh factor display
SET

Conditions：
At manual reset setting
RESET/SHIFT

Instant measurement display(General display)
MAX/MIN
Changes a screenwith the main
monitor factor
of instantmeasurement.

Push
1 seconds

Maximum / Minimum
measurement display

Maximum / Minimum
reset(Main monitor
display factor)＋＋
RESET/SHIFTPush 1 seconds

No operation,
10 minute
DISPLAY

Conditions：
Displaying the factor
applicable to themain monitor.

It displays thefactor of the
maximum and theminimum
measurement on
display on themain monitor.
MAX/MIN

Alarm reset(Output and display)

A,DA,MDA,V factor display.
At 1φ2W, With no function.

Phase (line) display
changeDISPLAY

Push 3 seconds or
No operation, 10 minute.
＋ , －DISPLAY

Maximum / Minimum
reset(All factor)

Electric-energy
integrated-value
enlarged display

Main monitor
display-elementchangePush 3 seconds or
No operation, 10 minute.
RESET/SHIFT＋－

Setting point check
DISPLAYOr No operation, 10 minute.
SET

Push 3 seconds
Setting mode 1
DISPLAYOr No operation, 10 minute.
SET＋
RESET/SHIFTPush 3 seconds
Setting mode 2
DISPLAYOr No operation, 10 minute.
SET＋
DISPLAY

Push 3 seconds
Setting mode 3
DISPLAYOr No operation, 10 minute.
SET
＋
MODE

Push 3 seconds
TEST mode
DISPLAY

11

SFLC-212-173

4.2 The kind of display
4.2.1 Measurement display
The change of the measurement display element of the main monitor by switch operation and the change ofthe phase / line display of current / voltage is possible (temporarily).
In a general measurement display, if switch operation is not performed for 10 minutes after changing adisplay element, it returns to the original measurement display element automatically.
Measurement factors, such as current, voltage,
and active power, are displayed.
The measurement value of four factors is
displayed.
Setting which always displays a measurement
factor is possible. And, it is possible tochange to a display of the maximum value and theminimum value, about the measurement factor
which performs holding of the maximum value andthe minimum value by switch operation.
These maximum values and the minimum value arereset by switch operation (it updates to theinstantaneous value at the time).
In addition, as for the maximum value and theminimum value, power-supply reset is notcleared either. And, this display is held byeven after 10 minutes of switch non-operation.

A(V) display

Instant display

V(UV) display

Wh display

Range of minimum to maximum isdisplayed in bar graph.

Minimum display
Maximum display

W display

General measurement display

General measurement
maximum and minimum display

・The example of a measurement display of each measurement factor (Main monitor)
Measurement
factorExample of display

Measurement
factor

Note“DEMAND”
display

Active power
W

Maximum
demand power
MDW

Note

Current
A

VoltageV

Demandcurrent
DA

Example of display“MAX
DEMAND”
display

12

Maximum
demandcurrent
MDA“MAX
DEMAND”
display

Demand power
DW“DEMAND”
display

Reactive
power
var

LAG
or
LEAD
displaySFLC-212-173

Measurement
factorExample of display

Measurement
factor

Note

LAG
or
LEAD
display

Power factor
cosφ

Example of display

Note

Frequency
Hz

Watt-hour(Powerreceiving)Wh

Watt-hour

var-hour(Powerreceiving，
LAG)
varh(LAG)“LAG”
display

var-hour(Powerreceiving，
LEAD)
varh(LEAD)

var-hour(Powertransmission,
LAG)-varh(LAG)“LAG” and”－(minus)”
display“－(minus)”display(Powertransmission)-Wh“LEAD”
display

var-hour(Powertransmission,
LEAD)“LEAD” and”－(minus)”
display-varh(LEAD)

4.2.2 Alarm detection display
The alarm value setting is a possible measurement factor (demand current and demand power, etc.), it
displays in case an input exceeds a set point.
Besides the usual measurement display, the detected factor is displayed on a screen upper case.
In addition, in case setting OFF (not use) as measurement factor, it does not detect.
And in the case of with an alarm-output option, it is possible to do an alarm output (relay make contact) tothe outside simultaneously with a screen display.
Alarm display possible factor) Demand current, Demand power, Voltage

Alarm detection(Demand current upper limit )A(V) display
Alarm
detection
W display

V(UV) display
Wh display
General measurement display.

General measurement display(Alarm detection)

13

SFLC-212-173

・The example of a display at the case of the detection in each alarm factor.
In case the alarm factor is indicating by measurement at the main monitor or the sub monitor, a measurement
value constitutes a blinking display.
The displays after an alarm return.
In case a return method is automatic reset setting.
It returns to the usual measurement display.
In case a return method is manual reset setting.
A detection display and an alarm output hold (in case setting as an alarm output of applicable factor).
The return in this case needs alarm reset operation. Please refer to "4.3.7 Reset" about alarm reset.

Alarm factor

Demandcurrent

Example of a display

Alarm factor

Upper
limit
(H)

Demand
power

Upper
limit
(H)
Alarm setting value
Detection display(At alarm factor setting)

Alarm setting value
Detection display(At alarm factor setting)

Voltage

Upper
limit
(H)

Example of a display

Voltage
Low-alarmsetting value
High-alarm
setting value

Lower
limit
(L)

Low-alarmsetting value
High-alarm
setting value

Detection display(At alarm factor setting)

Detection display(At alarm factor setting)14

SFLC-212-173

4.2.3 Setting display
It is the display at the case of various setting. There are three types of setting modes according to thecontents of a setting.
Operation and the contents of setting (detail) in setting mode, please refer to "5 Setting".
① Setting mode 1
Setting of a measurement display element, an
alarm output, and an alarm value is mainly
performed.
And, an alarm output can be tested in thissetting mode.

New setting
Current setting
Setting No.

Setting mode 1(No.111 Display pattern)② Setting mode 2
Setting of measurement range, analog output,
pulse output, and measurement display ON/OFF
is mainly performed.
And, initialization of a setting value andreset of watt-hour integrated value can be
performed in this setting mode.

New setting
Current setting
Setting No.
Setting factor
Setting mode 2(No.211 Voltage range)③ Setting mode 3
Setting of an input circuit and tidal current
measurement is mainly performed.
And, analog output can be adjusted in thissetting mode.

New setting
Current setting
Setting No.

Setting mode 3(No.311 Input circuit phase line change)

15

SFLC-212-173

4.3 Operation
4.3.1 The main monitor display-element change
The measurement display element of the main monitor is changed. A change is performed by ＋ －.
A measurement display and maximum display，minimum display can also perform this operation.
After changing a measurement display element, if a switch is not operated for 10 minutes, it will returnto the original measurement display element automatically.
In a maximum display and minimum display, even if a switch is not operated for 10 minutes, it does not
return to the original display.
A(V) display

No operation 10 minute

－

－

＋

＋

＋

＋

－

－

V(UV)
display

Main monitor A(U)W
displayWhdisplay

Main monitor A(V)
(Initial display)

Main monitor A(W)

4.3.2 Phase (line) display change (Three-phase three-wire, Single-phase three-wire)
A phase (line) display of current or voltage is changed. (Everything which is being indicated.)
A change is performed by DISPLAY .
A measurement display and maximum display，minimum display can also perform this operation.
In addition, after changing a phase (line) display, if a switch is not operated for 10 minutes, it will returnto the original phase (line) display automatically.
In a maximum display and minimum display, even if a switch is not operated for 10 minutes, it does not
return to the original display.
A(V) display

No operation
10 minute

DISPLAY

V(UV)
display

DISPLAYWh
W
display display

16

DISPLAYSFLC-212-173

4.3.3 Enlarged display of integral power consumption.
In case electric energy is being displayed by the general measurement display, an electric-energy display
is changed to a normal display (5 digits of integers), and an enlarged display (2 digits integer ＋ below decimal
point, 3 digits). A change is performed by SET . After an enlarged display, if a switch is not operated for
10 minutes, it returns to the usual display automatically.
<Caution > If it continues pushing SET 3 seconds or more, it will become the setting mode 1.
A(V) display

No operation 10 minute

SET

SET

Watt-hour enlarged display

W display Wh display
V(UV) display
Normal display

Enlarged display

4.3.4 Setting value check
A voltage range (primary voltage), a current range (primary current), and an alarm-output set point arechecked. Check is RESET/SHIFT and － are pushed simultaneously and performed.
The change of a set point is carries out by ＋ and － . DISPLAY is pushed in case it returns to theoriginal measurement display. And, if a switch is not operated for 10 seconds after a set point check,it will return to the original measurement display automatically.
A(V) display

No operation
10 minutePrimary voltage

RESET/SHIFT
＋
－

＋

－
DISPLAY

V(UV) display
Wh display
W display
Measurement display

Factor display
Input-rating voltage display(At 3φ3W，1φ2W)
Set point check
Voltage range (primary voltage),
Rated voltage－＋

Time interval，Contact delay
time

Alarm factor，Upper limit(lower limit)
value
Return method
(７)

－

＋
７

( )

Note(７) It is not displayed if there is no option.
17

－

＋

Current range(primary
current)

SFLC-212-173

4.3.5 Setting mode
Various kinds of setting are performed. Setting mode is three types, and operations are different.
DISPLAY is pushed in case it returns to the original measurement display. And, if a switch is not operated
for 10 minute after a set point check, it will return to the original measurement display automatically.
Operation and the contents of setting (detail) in setting mode, please refer to "5 Setting".
・Setting mode 1：Press SET for longer than 3 seconds.
・Setting mode 2：Press SET and RESET/SHIFT together for longer than 3 seconds.
・Setting mode 3：Press SET and DISPLAY together for longer than 3 seconds.
<Reference> A measurement display and maximum display，minimum display can also perform this operation.
A(V) display

Setting No.

New setting

Current setting

SET 3 seconds
SET ＋ RESET/SHIFT 3 seconds
SET ＋ DISPLAY 3 seconds

V(UV) display

Setting mode 1

Wh display

W display

Setting factor
Current setting
New setting

Measurement display

DISPLAY
Return tomeasurement
display
Setting No.
Setting mode 2
Setting No.
Current setting
New setting

Setting mode 3

18

SFLC-212-173

4.3.6 Reset
Various kinds of reset are performed. The kind of reset is as follows and operations are different,
respectively. Reset of integral power consumption (zero clear)， Reset of maximum value and minimum value(it updates to the instantaneous value at the time)， Alarm-output reset (OFF of an alarm output (at thecase of manual reset setting)).
And, the operation from each measurement display constitutes conditions at each reset.
(1) Integral power consumption reset
It resets by package about the integrated value of various watt-hour. Watt-hour reset is performed in
the setting mode 2. In detail explanation in the setting mode 2, please refer to "5.3.2 Setting mode 2".
① Press SET and RESET/SHIFT together for longer than 3 seconds.
② MODE is pushed until setting is set to No.261.

It goes into the setting mode 2.

Further RESET/SHIFT is pushed once and it is made awatt-hour-reset display by No.262.
③ Press SET for longer than 3 seconds.
④ DISPLAY is pushed and it returns to a measurement display.
A(V) display

New setting

SET ＋ RESET/SHIFT
3 seconds
DISPLAY

V(UV) display

Wh display

Setting factor
Current setting
Setting No.

W display
Measurement display

No.211
MODE

Setting No.

It becomes the "CLEAr"
display after "0" displays.

RESET/SHIFT×ｎ(To setting No.262)

SET 3 seconds

Wh and varh,
alternatelylights

Wh and varh,
alternately-l
ights

Setting No.

No.262（Clear completion）

It displays by
clear completion.
Setting mode 2

19

No.262（Before clear）

SFLC-212-173(2) Alarm reset
In case an alarm return method is set to “HOLD (manual return)”, an alarm output is reset (output OFF).
(With an alarm-output option)
However, an output is not turned off by this operation, in case an alarm continues and it has caused.
And, this operation is unnecessary if an alarm return method is set as "AUTO (automatic return)".
(An output is also OFF to an alarm return.)
In addition, setting can perform same operation in external operation input.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method.
Please refer to "6.3 Option" for the external operation input.
① It continues pushing RESET/SHIFT 1 second or more by a measurement display, the maximum measurement
display, and the minimum measurement display.
<Caution> If a RESET/SHIFT switch performs alarm reset, the maximum value of the measurement factor
currently displayed on the main monitor and the minimum value are also reset.
Alarm detection (Hold)

A(V) display

RESET/SHIFT
1 second

Instant measurement display

V(UV) display W display Wh display
Reset completion(3) Maximum value and minimum value reset.
Reset of the various measurement values of maximum value and minimum value is performed.
This reset has two types of methods. (How to perform according to a measurement factor individual.
How to reset all maximum values and minimum values by package.)a) Individual reset
It resets the differential maximum value and minimum value.
are not reset by this operation.

Other maximum values and minimum values① It displays a measurement factor to reset on the main monitor. (By measurement display and the
maximum and the minimum measurement display.)② Press RESET/SHIFT for longer than 1 seconds.
<Caution> Please be sure to perform this operation after displaying the maximum value and a minimum
value measurement factor to reset. And, if the maximum value and minimum value reset are
performed, the alarm output of detection will also be reset.
Instant display

It displays in bar graph, therange of minimum to maximum.

RESET/SHIFT
1 second

Minimum display

Maximum display

Minimum display(Reset)Maximum, Minimum measurement display
20

Instant display
Reset completion

Maximum display(Reset)SFLC-212-173b) All reset of maximum value and minimum value.
It resets all the maximum values and minimum values.
In addition, setting can perform same operation in external operation input.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method.
Please refer to "6.3 Option" for the external operation input.
① By measurement display and the maximum and minimum measurement display, it continues pushing
RESET/SHIFT and ＋ 1 second or more simultaneously.
Instant display

It displays in bar graph, therange of minimum to maximum.

RESET/SHIFT ＋ ＋
1 second

Minimum display

Maximum display

Minimum display(Reset)Maximum, Minimum measurement display

Instant display

Maximum display(Reset)Reset completion(4) Alarm reset, and all reset of maximum value and minimum value.
It resets an alarm in an external operation input. And, it resets all the maximum values and minimum
values.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method.
Please refer to "6.3 Option" for the external operation input.

21

SFLC-212-173

5. Setting< Caution >When changing the input circuit setting, please be sure to perform a setup from an input circuit settingin the setting mode 3. After changing the other setting, when the input circuit setting is changed theset value returns to default value (default value of a changed input circuit).
5.1 Function table
This product has each function setting with a front switch.
<Caution> In case the input circuit is not designated at the case of an order, it is shipping in the default
value of 3-phase 3-wire 110V input.
Setting mode 1.
Setting
No.

Function table
Function

111

Display pattern

Sets the display combination patternof the digital 4 displays and bar graphdisplay.

112

Main monitor

Sets the display factor of digital main
monitor.

113

sub monitor (left)

Sets the display factor of digital sub
monitor (left).

114

sub monitor (center)115

sub monitor (right)

116
121AL
(８)
122AL
(８)
123AL
(８)
131H
132
133H
134
135
141H
142L
151
152

Importantsetting

Page

○

34-38

○

34-38

○

34-38

W

○

34-38

Wh

○

34-38

○

34-38

○

38

Functional description

Sets the display factor of digital sub
monitor (center).
Sets the display factor of digital sub
monitor (right).

Default setting
Pattern 1
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

A(V)
A(U)
A
V(UV)
V(UN)
V

Bar graph

Sets the display factor of bar graph.

3φ3W
1φ3W
1φ2W

A(V)
A(U)
A

Alarm factor

Sets the output factor of alarm.

DA

Alarm return method

Sets the output action at the case ofan alarm return.

Automatic reset

38

Alarm contact delay time

Sets the contact delay time of alarm.

0 second

38

Demand current upper
limit
Demand current timeinterval
Demand power upper limit
Demand power timeinterval

Sets the high-alarm value of demandcurrent.

80% (Full scale＝
100%)

39

Sets time interval of demand current.

0 second

Sets the high-alarm value of demand power.

OFF (No operation)

Sets the time interval of demand power.

0 second○

39

Demand power operating
methodSets the operating method of demand
active power.

Operating method
according withbimetal type.

○39

Instant measurement
voltage upper limit
Instant measurement
voltage lower limit
Backlight action
Backlight brightness

Sets the high-alarm value of instantvoltage.
Sets the low-alarm value of instantvoltage.
Sets the ON/OFF of backlight.
Sets the brightness of backlight.

39
39

OFF (No operation)

40

OFF (No operation)

40

AUTO OFF3（Middle）

40
40

Note(８) A setting item is not displayed in case there is no corresponding option.

22○

SFLC-212-173

Setting mode 2.
Setting
No.

Function table (1)Function

Functional description

211

Voltage range

Sets the voltage-measurement range(primary voltage).

212

Digit number of voltage
range

Sets the digit number of voltage range.

213

Current range

Sets the current-measurement range(primary current).

214

Current display
intrinsic sensitivity

Sets the full scale of current meter.

215

Digit number of current
range

Sets the digit number of current range.

216

Active power polarity

Sets the deflection display of active
power meter.

217

Active power range

Sets the full scale of active power
meter.

218

Digit number of active
power range

Sets the digit number of active power
range.

219

Reactive power range

Sets the full scale of reactive power
meter.

21A

Digit number of reactive
power range

Sets the digit number of reactive
power range.

21B

Power-factor range

21C

Frequency range

21D

Digit number of
frequency range

221A
(９)

AO1 output factor

Sets the output factor of AO1 (analog
output 1).

222A
(９)

AO2 output factor

Sets the output factor of AO2 (analog
output 2).

223A
(９)

AO3 output factor

224A
(９)

Current output intrinsic
sensitivity

Sets the full scale of power-factor
meter. And, sets the output range of
analog output.
Sets the full scale of frequency meter.
And, sets the output range of analog
output.
Sets the digit number of frequency
range.

Sets the output factor of AO3 (analog
output 3).
Sets the analog output sensitivity (％of a rated input power value to an output
upper limit) of current.

Default setting
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

Page

○

42-45

10

6600V
( )
110.0V3300V(10)
4 digits
4 digits
4 digits
100.0A
500.0A
50.00A
100.0A
500.0A
50.0A
4 digits
4 digits
4 digits

42-45

○

3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

42-45

42-45

42-45

One-way deflection

42-45

11

1200kW( )100.0kW150.0kW (12)4 digits
4 digits
4 digits
600.0kvar (11)50.00kvar
75.00kvar (12)4 digits
4 digits
4 digits

42-45

42-45

42-45

42-45

0.5 to 1 to 0.5

42-45

45.0 to 65.0Hz

42-45

3 digits

42-45

3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

A(V)
A(U)
A
V(UV)
V(UN)
V

W
100.0％

Note(９) A setting item is not displayed in case there is no corresponding option.
Note(10) It is set to “220.0V (4-digits)” in 220V input.
Note(11) It is set to "40.00kW (4-digits)" and "20.00kvar (4-digits)" in 220V input.
Note(12) It is set to "10.00kW (4-digits)" and "5.000kvar (4-digits)" in 220V input.

23

Importantsetting

○

46

○

46

○

46
46

SFLC-212-173

Setting mode 2.
Setting
No.

Function table (2)Function

225A
(13)

Active power output
intrinsic sensitivity

226A
(13)

Reactive power output
intrinsic sensitivity

227A
(13)

Low input cut

231P
(13)

Pulse output factor

232P
Pulse unit
(13)(14)241(13)

External operation input
function

251

Voltage ON/OFF

252

Current ON/OFF

253

Demand current ON/OFF

254

Active power ON/OFF

255

Demand power ON/OFF

256

Reactive power ON/OFF

257

Power factor ON/OFF

258

Frequency ON/OFF

259
25A

Power receiving watthour ON/OFF
Power transmission
var-hour ON/OFF

25B

Power receiving varhour (LAG,LEAD) ON/OFF

25C

Power transmission
var-hour (LAG,LEAD)
ON/OFF261

set value initialization

262

Watt-hour reset

Functional description
Sets the analog output sensitivity (％of a rated input power value to an output
upper limit) of active power.
Sets the analog output sensitivity (％of a rated input power value to an output
upper limit) of reactive power.
Sets the function which makes a lower
limit the analog output at the case of aminute input (adequate to 0.5％ or less)in analog output.

Default setting

46

50.0％

46

OFF (No operation)

46

Wh

Sets the pulse unit of pulse output.

3φ3W
1φ3W
1φ2W

○

47

10kWh/pulse
47

1kWh/pulse

Alarm reset

○

47

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

ON

48

－

48

－

48

Note(13) A setting item is not displayed in case there is no corresponding option.
Note(14) It is set to “0.1kWh/pulse” in 220V input. (3φ3W, 1φ2W)

24

Page

100.0％

Sets the factor of pulse output.

Sets the function of the external
operation input.
Sets the ON/OFF of voltage measurement
display.
Sets the ON/OFF of current measurement
display.
Sets the ON/OFF of demand current
measurement display.
Sets the ON/OFF of active power
measurement display.
Sets the ON/OFF of demand power
measurement display.
Sets the ON/OFF of reactive power
measurement display.
Sets the ON/OFF of power-factor
measurement display.
Sets the ON/OFF of frequency measurement
display.
Sets the ON/OFF of power-receiving
watt-hour measurement display.
Sets the ON/OFF of power transmission
var-hour measurement display.
Sets the ON/OFF of power-receiving
var-hour (LAG,LEAD) measurement
display.
Sets the ON/OFF of power transmission
var-hour (LAG,LEAD) measurement
display.
Initialize the settings of setting 1 andsetting 2 (return to the default
settings).
(The set value of setting 3 does not
return to the default value)
Integrated value is cleared by package
about each electric energy.

ImportantsettingSFLC-212-173

Setting mode 3.

Function table

Setting
No.

Function

Functional description

311

Input circuit phase line
changeSets the input circuit or phase line.

312

Input voltage

Sets the input voltage or phase voltage
full scale.

321

331
(15)
332
(15)
333
(15)
334
(15)
335
(15)
336
(15)

Tidal current measurement

AO1 BIAS adjustment
AO1 SPAN adjustment
AO2 BIAS adjustment
AO2 SPAN adjustment
AO3 BIAS adjustment
AO3 SPAN adjustment

Sets the general measurement or tidal
current measurement which was consciousof power transmission / power receiving,in measurement of reactive power andpower-factor.
Sets the BIAS value of AO1 (analog
output 1).
Sets the SPAN value of AO1 (analog
output 1).
Sets the BIAS value of AO2 (analog
output 2).
Sets the SPAN value of AO2 (analog
output 2).
Sets the BIAS value of AO3 (analog
output 3).
Sets the SPAN value of AO3 (analog
output 3).

Default setting
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

3φ3W
1φ3W(U-W-N)
1φ2W
110V
150V
110V

Page

○

50

○

50

General measurement

51

0.0％

52

100.0％

52

0.0％

52

100.0％

52

0.0％

52

100.0％

52

Note(15) A setting item is not displayed in case there is no corresponding option.

25

ImportantsettingSFLC-212-173

5.2 Setting table
A setting item changes by the specification of a product, or the existence of an option.
(1) Important setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Sets the phase wireof input circuit.
(311)

Setting and operation procedures

Page

Press SET and DISPLAY together for longer than 3 seconds(311)
Press SET for longer than 3 seconds
Select an phase wire by ＋ and －Selected phase wire is entered

Press DISPLAY

50

Returns to display mode.

Press SET and DISPLAY together for longer than 3 seconds
Sets the input
voltage(phase-voltage
full scale).
(312)

Press RESET/SHIFT
(311)
(312)
Select an input voltage (In case of 1φ3W, it is phase-voltage full scale) by ＋ and －50
Press SET

Selected input voltage is entered

Press DISPLAY

Returns to display mode.
Sets themeasurement rangeof voltmeter(211)

Sets themeasurement rangeof ammeter(213)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Select a measurement range by ＋ and －Press SET
Selected measurement range is entered

Press DISPLAY

42-45

Returns to display mode.

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Select a measuring range by ＋ and －
(212)
(213)
Press SET
Selected measuring range is entered
Press DISPLAY

42-45

Returns to display mode.
Press SET for longer than 3 seconds
Sets the display
combination(111)

Press SET

Select the display combination by ＋ and －(111)
Selected display combination is entered
Press DISPLAY

34-38

Returns to display mode.
Sets the output
factor of analog
output 1 (AO1).
(221A)

Sets the output
factor of analog
output 2 (AO2).
(222A)

Sets the output
factor of analog
output 3 (AO3).
(223A)

Press SET and RESET/SHIFT together for longer than 3 seconds
Press MODE
(211)
(221A)
Press SET
Select an output factor by ＋ and －Selected output factor is entered

Press DISPLAY

Returns to display mode.

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Select an output factor by ＋ and －Press SET(222A)
Selected output factor is entered
Press DISPLAY
Returns to display mode.
Press SET and RESET/SHIFT together for longer than 3 seconds
Press MODE
(211)
(221A)
Press RESET/SHIFT
Select an output factor by ＋ and －Press RESET/SHIFT(222A)(223A)
Press SET
Selected output factor is entered
Press DISPLAY
Returns to display mode.

26

46

46

46

SFLC-212-173

Items
Sets the output
factor of pulse
output.
(231P)

Sets the factor
of alarm output.
(121AL)

Setting and operation procedures

Page

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Select an output factor by ＋ and －Press SET
(231P)
Selected output factor is entered
Press DISPLAY
Returns to display mode.

47

Press SET for longer than 3 seconds
Press MODE
(111)
(121AL)
Press SET
Selected factor is entered
Select an factor by ＋ and －Press DISPLAY

38

Returns to display mode.

Sets the functionof external
operation input.
(241)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Select an function by ＋ and －Press SET
Press MODE
(231P)(241)
Selected function is entered
Press DISPLAY
Returns to display mode.

47

Sets the timeinterval of demandcurrent.
(132)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Select an time interval by ＋ and －Press SET
(132)
Press DISPLAY
Returns to display mode.
Selected time interval is entered

39

Sets the timeinterval of demand
power.
(134)

Press SET for longer than 3 seconds
Press MODE
Press MODE
(111)
(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)
Select an time interval by ＋ and －Press SET
Selected time interval is entered

Sets the operation
method of demand
power.
(135)

Press DISPLAY

Returns to display mode.

Press SET for longer than 3 seconds
Press MODE
Press MODE
(111)
(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)
Press RESET/SHIFT
Select an operation method by ＋ and －Press SET
(135)
Press DISPLAY
Returns to display mode.
Selected operation method is entered

27

39

39

SFLC-212-173(2) A combination except a display pattern.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Sets the display
factor of main
monitor.
(112)

Setting and operation procedures

Page

Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Select an display factor by ＋ and －Press SET
Selected display factor is entered

Press DISPLAY

34-38

Returns to display mode.

Sets the display
factor of sub
monitor (left).
(113)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Select an display factor by ＋ and －Press SET
(113)
Selected display factor is entered
Press DISPLAY
Returns to display mode.

34-38

Sets the display
factor of sub
monitor (center).
(114)

Press SET for longer than 3 seconds
Press RESET/SHIFT
Press RESET/SHIFT
(111)
(112)
Press RESET/SHIFT
Select an display factor by ＋ and －Press SET
(113)
(114)
Selected display factor is entered
Press DISPLAY
Returns to display mode.

34-38

Sets the display
factor of sub
monitor (right).
(115)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Press RESET/SHIFT
Press RESET/SHIFT
(113)
(114)
(115)
Select an display factor by ＋ and －Press SET
Press DISPLAY

Selected display factor is entered

Sets the display
factor of bar
graph.
(116)

Returns to display mode.

Press SET for longer than 3 seconds
Press RESET/SHIFT
Press RESET/SHIFT
(111)
(112)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(113)
(114)
(115)
(116)
Select an display factor by ＋ and － (If a sub monitor is selected, an underbar
will be displayed on the bottom of a digital display.)
Press SET

Selected display factor is entered

34-38

Press DISPLAY

34-38

Returns todisplay mode.

(3) Setting of active power polarity and measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Sets the polarityand measurement
range of active
power.
(216)(217)

Setting and operation procedures

Page

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
Select an one-way deflection (P) / both deflection (-) by ＋ and －Press SET

Selected deflection is entered

Press RESET/SHIFT(217)

Select a measuring range by ＋ and －Press SET

Selected measuring range is entered

Press DISPLAY

28

Returns to display mode.

42-45

SFLC-212-173(4) Setting of reactive power measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Sets themeasurement rangeof reactive power.
(219)

Setting and operation procedures
Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
(217)
Press RESET/SHIFT
Select a measuring range by ＋ and －Press RESET/SHIFT
(218)
(219)
Press SET
Selected measuring range is entered
Press DISPLAY

Page

42-45

Returns to display mode.

(5) Setting of Wh (varh) output pulse unit.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Setting and operation procedures

Sets the pulse unitof pulse output.
(232P)

Press SET and RESET/SHIFT together for longer than 3 seconds
Press MODE
(211)
(221A)
Press MODE
Press RESET/SHIFT
Select a output pulse unit by ＋ and －(231P)(232P)
Selected output pulse unit is entered
Press DISPLAY
Press SET

Page

47

Returns to display mode.

(6) Setting of power-factor and frequency measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Setting and operation procedures

Page

Sets themeasurement rangeof power-factor.
(21B)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
(217)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(218)
(219)
(21A)
Press RESET/SHIFT
Select a measuring range by ＋ and －Press SET
(21B)
Press DISPLAY
Returns to display mode.
Selected measuring range is entered

42-45

Sets themeasurement rangeof frequency.
(21C)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
(217)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(218)
(219)
(21A)
Press RESET/SHIFT
Select a measuring range by ＋ and －Press RESET/SHIFT
(21B)
(21C)
Press SET
Selected measuring range is entered
Press DISPLAY
Returns to display mode.

29

42-45

SFLC-212-173(7) Setting of current display intrinsic sensitivity.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Sets the display
sensitivity (％ ofa display to an
input) of current.
(214)

Setting and operation procedures

Page

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Select a display sensitivity by ＋ and －Press SET
Selected display sensitivity is entered

Press DISPLAY

42-45

Returns to display mode.

(8) Setting of range digit number.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Setting and operation procedures

Page

Sets the digit
number of voltage
range.
(212)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Select a digit number by ＋ and －Press SET
(212)
Selected digit number is entered
Press DISPLAY
Returns to display mode.

42-45

Sets the digit
number of current
range.
(215)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Select a digit number by ＋ and －Press SET
(215)
Selected digit number is entered
Press DISPLAY
Returns to display mode.

42-45

Sets the digit
number of active
power range.
(218)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
(217)
Press RESET/SHIFT
Select a digit number by ＋ and －Press SET
(218)
Press DISPLAY
Returns to display mode.
Selected digit number is entered

42-45

Sets the digit
number of reactive
power range.
(21A)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
(217)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(218)
(219)
(21A)
Select a digit number by ＋ and －Press SET
Press DISPLAY

Selected digit number is entered

30

Returns to display mode.

42-45

SFLC-212-173

Items

Sets the digit
number of
frequency range.
(21D)

Setting and operation procedures

Page

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(212)
(213)
(214)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(215)
(216)
(217)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(218)
(219)
(21A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(21B)
(21C)
(21D)
Select a digit number by ＋ and －Press SET
Press DISPLAY

Selected digit number is entered

42-45

Returns to display mode.

(9) Setting of analog output.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Setting and operation procedures

Sets the analog
output sensitivity(％ of a rated
input power value
to an output upper
limit) of current.
(224A)

Press SET and RESET/SHIFT together for longer than 3 seconds
Press MODE
(211)
(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)
Select a output sensitivity by ＋ and －Press SET

Sets the analog
output sensitivity(％ of a rated
input power value
to an output upper
limit) of active
power.
(225A)
Sets the analog
output sensitivity(％ of a rated
input power value
to an output upper
limit) of reactive
power.
(226A)

Press SET and RESET/SHIFT together for longer than 3 seconds
Press MODE
(211)
(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)
Select a output sensitivity by ＋ and －Press SET
(225A)
Press DISPLAY
Returns to display mode.
Selected output sensitivity is entered

46

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)
Press RESET/SHIFT
Select a output sensitivity by ＋ and －Press SET
(225A)(226A)
Selected output sensitivity is entered
Press DISPLAY
Returns to display mode.

46

Sets the output cut
function at thecase in minute
input (0.5％ orless) of analog
output.
(227A)

Selected output sensitivity is entered

Press DISPLAY

Page

Returns to display mode.

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)
Press RESET/SHIFT
Press RESET/SHIFT(225A)(226A)(227A)
Press SET
Select a low Input cut ON/OFF by ＋ and －Selected action is entered

Press DISPLAY

3146

Returns to display mode.

46

SFLC-212-173(10) Setting of alarm output.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Sets the return
method of alarm
output.
(122AL)

Sets the contact
delay time ofalarm output.
(123AL)

Setting and operation procedures

Page

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Select a return method by ＋ and －Press SETThe selected return method is entered

Press DISPLAY

38

Returns to display mode.

Press SET for longer than 3 seconds
Press MODE
Press RESET/SHIFT(111)(121AL)(122AL)
Press RESET/SHIFT
Select an contact delay time by ＋ and －Press SET
(123AL)The selected contact delay time is entered
Press DISPLAY
Returns todisplay mode.

38(11) Demand measurement (current, active power) setting.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Sets the high-alarm
value of demandcurrent.
(131H)

Sets the high-alarm
value of demand
power.
(133H)

Setting and operation procedures

Page

Press SET for longer than 3 seconds
Press MODE
Press MODE
(111)
(121AL)(131H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

39

Returns to display mode.

Press SET for longer than 3 seconds
Press MODE
Press MODE
(111)
(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Select a high-alarm value by ＋ and －(132)(133H)
Press SET
Selected high-alarm value is entered
Press DISPLAY

39

Returns to display mode.

(12) Instant measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Setting and operation procedures

Page

Sets the high-alarm
value of instantvoltage.
(141H)

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press MODE
Select a high-alarm value by ＋ and －Press SET(141H)
Press DISPLAY
Returns to display mode.
Selected high-alarm value is entered

40

Sets the low-alarm
value of instantvoltage.
(142L)

Press SET for longer than 3 seconds
Press MODE
Press MODE
(111)
(121AL)(131H)
Press MODE
Press RESET/SHIFT
Select a low-alarm value by ＋ and －(141H)(142L)
Press SET
Selected low-alarm value is entered
Press DISPLAY
Returns to display mode.

32

40

SFLC-212-173(13) Backlight setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items

Setting and operation procedures

Page

Sets the action of
backlight.
(151)

Press SET for longer than 3 seconds
Press MODE
Press MODE
Press MODE
(111)
(121AL)(131H)
Press MODE
Select a backlight action by ＋ and －Press SET(141H)(151)
Selected backlight action is entered
Press DISPLAY
Returns to display mode.

40

Sets the brightnessof backlight.
(152)

Press SET for longer than 3 seconds
Press MODE
Press MODE
Press MODE
(111)
(121AL)(131H)
Press MODE
Press RESET/SHIFT
Select a brightness by ＋ and －(141H)(151)(152)
Press SET
Selected backlight brightness is entered
Press DISPLAY

40

Returns to display mode.

(14) Other, measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Sets the tidal
current
measurement ofreactive power andpower-factor.
(321)

5.3

Setting and operation procedures

Page

Press SET and DISPLAY together for longer than 3 seconds
Press MODE
(311)
(321)
Press RESET/SHIFT
Select a tidal current measurement ON/OFF by ＋ and －(322)
Press SET
Selected action is entered
Press DISPLAY
Returns to display mode.

51

5.3.1

Setting in detail explanation
Setting mode 1SET
3 seconds
DISPLAY

111 to 116
Display setting
MODE

DISPLAY

121AL to 123AL
Alarm output setting
MODE

Display mode

DISPLAY

131H to 135
Demand detection setting
MODE

DISPLAY

141H to 142L
Instant measurement detection setting
MODE

DISPLAY

151, 152
Backlight setting
MODE

Setting mode 1

Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
33

SFLC-212-173(1) 111 to 116

Display combination setting● 3-phase 3-wire№

Pattern №

Main monitor

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Pattern 1
Pattern 2
Pattern 3
Pattern 4
Pattern 5
Pattern 6
Pattern 7
Pattern 8
Pattern 9
Pattern 10
Pattern 11
Pattern 12
Pattern 13
Pattern 14
Pattern 15

A(V)
A(V)
A(V)
DA(V)
MDA(V)
W
W
W
DW
MDW
A(V)
A(V)
W
A(V)
V(UV)

Sub monitor(Left)
V(UV)
V(UV)
V(UV)
V(UV)
A(V)
V(UV)
V(UV)
V(UV)
V(UV)
W
cosφ
var
cosφ
A(U)
V(VW)

Sub monitor(Center)W
W
W
MDA(V)
V(UV)
A(V)
A(V)
A(V)
MDW
V(UV)
W
W
var
A(W)
V(WU)

Sub monitor(Right)
Wh
cosφ
Hz
Wh
Wh
Wh
cosφ
Hz
Wh
Wh
Wh
Wh
Wh
Wh
Hz

Bar graph
A(V)
A(V)
A(V)
MDA＋DA(V)MDA＋DA(V)W
W
W
MDW＋DW
MDW＋DW
A(V)
A(V)
W
A(V)
V(UV)

● Displays set factor (3-phase 3-wire)
Main monitor

V(UV)，V(VW)，V(WU)，A(U)，A(V)，A(W)，DA(U)，DA(V)，DA(W)，MDA(U)，MDA(V)，MDA(W)，W，
DW，MDW，var，cosφ，Hz，Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)

Sub monitor(Left)

V(UV)，V(VW)，V(WU)，A(U)，A(V)，A(W)，W，var，cosφ

Sub monitor(Center)

V(UV)，V(VW)，V(WU)，A(U)，A(V)，A(W)，DA(U)，DA(V)，DA(W)，MDA(U)，MDA(V)，MDA(W)，W，
DW，MDW，var，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)

Sub monitor(Right)V(UV)，V(VW)，V(WU)，A(U)，A(V)，A(W)，DA(U)，DA(V)，DA(W)，MDA(U)，MDA(V)，MDA(W)，W，
DW，MDW，cosφ，Hz，Wh，-Wh

Bar graph

V(UV)，V(VW)，V(WU)，A(U)，A(V)，A(W)，DA(U)，DA(V)，DA(W)，MDA(U)，MDA(V)，MDA(W)，W，
DW，MDW, Var，cosφ，Hz● Phase (line) change (3-phase 3-wire) (16)
V(UV)

V(VW)

V(WU)

A(V)

A(W)

A(U)

Note(16) Press DISPLAY , Voltage and current are replaced at
the same time.
Note(17) DA and MDA also change.

(17)● Measurement factor change (3-phase 3-wire, Measurement display mode)
V(UV)

V(VW)

V(WU)

A(U)

A(V)

A(W)

DA(U)

DA(V)

DA(W)

MDA(U)MDA(V)

MDA(W)

W

DW

MDW

var

cosφ

Hz

Wh

-Wh

varh
(LAG)

varh
(LEAD)

-varh
(LAG)

-varh
(LEAD)

Nothing34

SFLC-212-173● Single-phase 3-wire (20)№

Pattern №

Main monitor

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Pattern 1
Pattern 2
Pattern 3
Pattern 4
Pattern 5
Pattern 6
Pattern 7
Pattern 8
Pattern 9
Pattern 10
Pattern 11
Pattern 12
Pattern 13
Pattern 14
Pattern 15

A(U)
A(U)
A(U)
DA(U)
MDA(U)
W
W
W
DW
MDW
A(U)
A(U)
W
A(U)
V(UN)

Sub monitor(Left)
V(UN)
V(UN)
V(UN)
V(UN)
A(U)
V(UN)
V(UN)
V(UN)
V(UN)
W
cosφ
var
cosφ
A(W)
V(WN)

Sub monitor(Center)
W
W
W
MDA(U)
V(UN)
A(U)
A(U)
A(U)
MDW
V(UN)
W
W
var
A(N)
V(UW)

Sub monitor(Right)
Wh
cosφ
Hz
Wh
Wh
Wh
cosφ
Hz
Wh
Wh
Wh
Wh
Wh
Wh
Hz

Bar graph
A(U)
A(U)
A(U)
MDA＋DA(U)MDA＋DA(U)W
WW
MDW＋DW
MDW＋DW
A(U)
A(U)
W
A(U)
V(UN)

● Displays set factor (Single-phase 3-wire) (20)Main monitor

V(UN)，V(WN)，V(UW)，A(U)，A(W)，A(N)，DA(U)，DA(W)，DA(N)，MDA(U)，MDA(W)，MDA(N)，W，
DW，MDW，var，cosφ，Hz，Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)

Sub monitor(Left)

V(UN)，V(WN)，V(UW)，A(U)，A(W)，A(N)，W，var，cosφ

Sub monitor(Center)

V(UN)，V(WN)，V(UW)，A(U)，A(W)，A(N)，DA(U)，DA(W)，DA(N)，MDA(U)，MDA(W)，MDA(N)，W，
DW，MDW，var，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)

Sub monitor(Right)V(UN)，V(WN)，V(UW)，A(U)，A(W)，A(N)，DA(U)，DA(W)，DA(N)，MDA(U)，MDA(W)，MDA(N) ，W，
DW，MDW，cosφ，Hz，Wh，-Wh

Bar graph

V(UN)，V(WN)，V(UW)，A(U)，A(W)，A(N)，DA(U)，DA(W)，DA(N)，MDA(U)，MDA(W)，MDA(N)，W，
DW，MDW，var，cosφ，Hz● Phase (line) change (Single-phase 3-wire) (18)
V(UN)

V(WN)

V(UW)

A(U)

A(W)

A(N)

Note(18) Press DISPLAY , Voltage and current are replaced at
the same time.
Note(19) DA and MDA also change.

(19)● Measurement factor change (Measurement display mode) (20)
V(UN)

V(WN)

V(UW)

A(U)

A(W)

A(N)

DA(U)

DA(W)

DA(N)

MDA(U)

MDA(W)

MDA(N)

W

DW

MDW

var

cosφ

Hz

Wh

-Wh

varh
(LAG)

varh
(LEAD)

-varh
(LAG)

-varh
(LEAD)

Nothing

Note(20) The case of single phase 3-wire (U-N-W).
The case of single phase 3-wire (U-N-V), voltage (UN-VN-UV) and current (U-N-V).
The case of single phase 3-wire (V-N-W), voltage (VN-WN-VW) and current (V-N-W).

35

SFLC-212-173● Single-phase 2-wire№

Pattern №

Main monitor

1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Pattern 1
Pattern 2
Pattern 3
Pattern 4
Pattern 5
Pattern 6
Pattern 7
Pattern 8
Pattern 9
Pattern 10
Pattern 11
Pattern 12
Pattern 13
Pattern 14
Pattern 15

A
A
A
DA
MDA
W
W
W
DW
MDW
A
A
W
A
V

Sub monitor(Left)
V
V
V
V
A
V
V
V
V
W
cosφ
var
cosφ
－
－

Sub monitor(Center)
W
W
W
MDA
V
A
A
A
MDW
V
W
W
var
－
－

Sub monitor(Right)
Wh
cosφ
Hz
Wh
Wh
Wh
cosφ
Hz
Wh
Wh
Wh
Wh
Wh
Wh
Hz

Bar graphA
A
A
MDA＋DA
MDA＋DAW
W
W
MDW＋DW
MDW＋DW
A
A
W
A
V● Displays set factor (Single-phase 2-wire)
Main monitor

V，A，DA ，MDA ，W，DW ，MDW ，var，cosφ，Hz，Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)

Sub monitor(Left)

V，A，W，var，cosφ

Sub monitor(Center)

V，A，DA ，MDA ，W，DW ，MDW ，var，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)

Sub monitor(Right)

V，A，DA ，MDA ，W，DW ，MDW ，cosφ，Hz，Wh，-Wh

Bar graph

V，A，DA ，MDA ，W，DW ，MDW ，var，cosφ，Hz● Measurement factor change (Measurement display mode)

V

A

DA

MDA

W

DW

MDW

var

cosφ

Hz

Wh

-Wh

varh
(LAG)

varh
(LEAD)

-varh
(LAG)

-varh
(LEAD)

Nothing36

SFLC-212-173
Display combination setting
MAX/MINRESET/SHIFT
111
Combination
display

RESET/SHIFT
MAX/MIN

112
Main monitor

Pattern 1

A(V)：5＋＋－RESET/SHIFT
MAX/MIN

113
Sub monitor(Left)RESET/SHIFT
MAX/MIN

V(UV)：1－＋

114
Sub monitor(Center)－＋－

A(W)：6

V(VW)：2

DW：14

＋

＋

＋

＋

－

＋

－

－

－

115
Sub monitor(Right)Main monitor

＋

－

－

＋

MDW：15

Nothing：0

＋

＋

＋

＋

－

A(U)：4

var：16

V(UV)：1

Pattern 14

＋

＋

＋

＋

＋

DA(W)：9－Pattern 15＋＋－

Manual setting
value：0－

A(V)：5＋－＋－

MDA(U)：10

A(W)：6

＋

＋

－

＋

var：8
＋

－

W：13
－

＋

＋

－

A(U)：4＋－－

A(V)：5＋－－

A(W)：6＋＋－

V(UV)：1

＋

＋

－

－

V(WU)：3

－

＋

－

－

cosφ：17

A(U)：4

＋

＋

－

－

Hz：18

A(V)：5

＋

＋

－

＋

－

－

－

DA(U)：7＋－

V(VW)：2

DA(V)：8＋－

DA(W)：9＋－＋－

MDA(V)：11
＋

－

Wh：19
＋

＋

＋

－

-Wh：20＋＋－varh(LAG)：21＋－

varh(LEAD)：22＋－-varh(LAG)：23＋－-varh(LEAD):24
＋

－

－

－

W：13
＋

－

Nothing：0

DA(V)：8

DW：14

＋

＋

＋

－

－

－

V(UV)：1

DA(W)：9

MDW：15＋＋＋－V(VW)：2＋－

cosφ：16

＋

＋

＋

－

－

MDA(U)：10－

MDA(V)：11

V(WU)：3＋－

At the time of
3-phase 3-wire.

MDA(U)：10

MDA(W)：12

DA(U)：7－－

A(W)：6

－

－

Hz：17

－

MDA(W)：12

A(U)：4

Default setting

37

－

－

Nothing：0＋＋

Sub monitor：Right－＋－

Sub monitor：Center－cosφ：9－

var：16＋＋－V(WU)：3-varh(LEAD):20－

MDW：15＋＋

Nothing：0

DW：14

＋

－

varh(LEAD)：18＋＋

V(VW)：2-varh(LAG)：19

＋

＋

＋

－

MDA(W)：12

＋

－

－

＋

－

－

varh(LAG)：17

W：7

MDA(V)：11
At the cases otherthan a display
combination pattern,
it becomes manual
setting.

－＋－

DA(V)：8－

Sub monitor
：Left

－

V(WU)：3－＋-Wh：19

DA(U)：7－

116
Bar graphfactorRESET/SHIFT
MAX/MIN

Wh：18

W：13

Pattern 2

Pattern 3

RESET/SHIFT
MAX/MIN－

SFLC-212-173◆ 111 Combination display
Select the factors to be measured and monitored by
4 digital displays out of combination patterns.
Set values are updated by SET .

New setting
Current setting◆ 112 to 115 Main monitor, Sub monitor (left),
Sub monitor (center), Sub monitor (right)
Sets these items for a display configuration otherthan combination patterns.
Set values are updated by SET .

Setting №

Combination display◆ 116 Bar graph factor
A factor being monitored in the main monitor isbasically displayed by a bar graph.
Set this item for displaying a factor being
monitored on a sub monitor by bar graph.
An underbar is attached to the digital displayof the setting sub monitor.
Set values are updated by SET .

Setting №

Underbar

Bar graph factor(2) 121AL to 123AL Alarm output setting 【With option】
Various setting and an output test are performed about an alarm output.

MAX/MIN

121AL
Alarm factor

MAX/MINRESET/SHIFTRESET/SHIFT

122AL
Alarm return method

MAX/MINRESET/SHIFT

123AL
Alarm contact delay time

◆ 121AL Alarm output factor setting
Sets the factor of alarms outputs. Selection by ＋ and － , set value is updated by SET .
Default setting：1 (DA：Demand current )OFF：Nothing＋

1：
Demand current－＋－

2：Demandactive power＋－

3：Voltage＋－◆ 122AL Alarm reset method setting.
Action at the case of a reset of alarm output can be selectedfrom AUTO (automatic reset) and HOLD (manual reset).
In "AUTO (automatic reset)", an alarm output is OFF according
to a reset of an alarm. In "HOLD (manual reset)", even afteran alarm reset, an output holds ON.
It performs a return (output OFF) in RESET/SHIFT .
Selection by ＋ and － , set value is updated by SET .
Default setting：AUTO (automatic reset)
AUTO(Automatic reset)

＋
－

＋

HOLD
(Manual reset)－

New setting

Setting №
Current setting
Alarm factor◆ 123AL Alarm contact delay time
Sets the contact delay time of alarm. The setting range is 0 to 300 seconds (1-second step).
Selection by ＋ and － , set value is updated by SET .
Default setting：0 second (With no contact delay)

38

SFLC-212-173(3) 131H to 135 Demand detection setting
The next operation method is setting. Demand current, Demand power, High-alarm value,
Time-interval, Power-factor, Demand power operation method.
131H
Demand current
upper limit

132
Demand current
time interval

MAX/MINRESET/SHIFT

133H
Demand power
upper limit

135
Demand power
operation method

RESET/SHIFT
MAX/MIN

134
Demand power
time interval

MAX/MINRESET/SHIFTRESET/SHIFT
MAX/MIN◆ 131H Demand current upper limit. 133H Demand power upper limit.
Sets the high-alarm value of demand current (DA) and demand power (DW).
The setting range is 5 to 100% (1% step) and OFF. (To full scale = 100%)
Selection by ＋ and － , set value is updated by SET .
Default setting：80％ (demand current), OFF (demand power)◆ 132 Demand current time interval. 134 Demand power time interval.
Sets the time interval (95％ time interval) of demand current(DA)and demand power(DW).
Selection by ＋ and － , set value is updated by SET .
Default setting：0 second (Demand current, Demand power)
0sec＋(Nothing)

－
－
＋

5sec

4min

5min
－
＋

30min

＋
－
－
＋
＋
－
－
＋

10sec

3min

6min

25min

＋
－
－
＋
＋
－
－
＋

20sec

2min

7min

20min

＋
－
－
＋
＋
－
－
＋

30sec

1min

8min

15min

＋
－
－
＋
＋
－
－
＋

40sec

MAX/MINRESET/SHIFT

New setting
Current setting

＋
－

50sec
＋

9min

－

10min

Setting factor
Setting No.
Demand current time interval◆ 135 Demand operation method
The operating system of demand (DW) can be selected
d：Operating systemfrom d (operating system according with bimetallic
according with
type：demand) and A (averaging arithmetic in demand
bimetallic typetime delay：average).
Selection by ＋ and － , set value is updated by SET .
Default setting：d (operation method according with bimetallic type)＋＋－

A：Averaging
arithmetic indemand time
delay

－

● Demand time delay characteristic (Demand current, Demand power)
100
95
80

Arithmetic method
according with bimetallic
type.
(Indication time to 95％
of a final constant value)

Time interval 10 minutes

Demand
value
60
（％）
40

20
0
0

10

20

30

Averaging arithmetic indemand time delay.
(Averaging time in demand
time delay.)Time（ min）

Arithmetic method
Demand current measurement：Arithmetic method according with bimetallic type.
Demand power measurement ：Arithmetic method according with bimetallic type (Default value).
Or the averaging arithmetic in a demand time interval.
One is selected by setting.
100％ indication time is about 3 times the time delay at the case of the arithmetic method according withbimetallic type.(In case of 10 minutes/95％ of time-interval, time to reach to 100％ is about 30 minutes.)
Demand measurement is measured to the 2 times of the rated current, and the 2 time of a rated active power.
39

SFLC-212-173(4) 141H to 142L Instant measurement detection setting
An upper limit low-alarm value is set by instant measurement (voltage factor).
141H
Instant measurement
voltage upper limit

142L
Instant measurement
voltage lower limit

MAX/MINRESET/SHIFT◆ 141H Instant measurement voltage upper limit,
142L Instant measurement voltage lower limit.

MAX/MIN

New setting

Current setting

RESET/
SHIFT

Setting No.

Sets the high-alarm value and low-alarm value of instantmeasurement (voltage full scale =150％).
Setting range is 30 to 150％ (1％ step).
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF〔Non-use〕(Upper limit, lower limit )Instant measurement
voltage upper limit(5) 151 to 152 Backlight setting
Sets the action and brightness of backlight.
151
Backlight action

MAX/MINRESET/SHIFT

152
Backlight brightness

MAX/MINRESET/SHIFT

New setting

Current setting◆ 151 Backlight action
It can select from ON (always-on), AUTO (auto off), andOFF (always-off) about action of backlight.
If 5 minutes elapses without operating a switch in case it isset as "AUTO (auto off)", backlight will go out automatically.
After that, backlight will be turned on if either of switches
is operated.
Selection by ＋ and － , set value is updated by SET .
Default setting：AUTO (Auto off)
AUTO：
Auto off

＋

OFF：

＋

ON：

－

Always-off－

Always-on－＋◆ 152 Backlight brightness
It can select the brightness of backlight as five steps of 1 to 5.
Backlight becomes the darkest if it is set as "1".
Backlight becomes the brightest if it is set as "5".
Selection by ＋ and － , set value is updated by SET .
Default setting：3 (Middle)

40

Setting No
Setting factor
Backlight

Setting54321Brightness
BrightDark

SFLC-212-173

5.3.2

Setting mode 2
SET andRESET/SHIFT
3 seconds
DISPLAY

211 to 21D
Measurement range setting
MODE

DISPLAY

221A to 227A
Analog output setting
MODE

DISPLAY

231P to 232P
Pulse output setting
MODE

Display mode
DISPLAY

241
External operation input setting
MODE

DISPLAY

251 to 25C
Measurement display ON/OFF setting
MODE

DISPLAY

261, 262
Other setting
MODE

Setting mode 2

Setting mode 2 is selected by pressing SET and RESET/SHIFT switches continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, a display and output of measurement are not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

41

SFLC-212-173(1) 211 to 21D Measurement range setting
Sets the measurement range of each measurement factor.
211
Voltagerange

MAX/MINRESET/SHIFTRESET/SHIFT
MAX/MINRESET/SHIFT
MAX/MIN

212
Digit numberof voltage
range
219
Reactive
power range
21A
Digit numberof reactive
power range

213

MAX/MIN

214
MAX/MINRESET/SHIFT

Current range

RESET/SHIFTRESET/SHIFT
MAX/MIN

218
Digit numberof active
power range

RESET/SHIFT
MAX/MIN

MAX/MINRESET/SHIFT

21B
Power-factor
range

MAX/MINRESET/SHIFT

Current display
intrinsic
sensitivity

MAX/MINRESET/SHIFT

215
Digit numberof current
range
RESET/

217
Active power
range

RESET/SHIFT
MAX/MIN

216
Active power
polarity

21C
Frequency
range

MAX/MINRESET/
SHIFT

SHIFT
MAX/
MIN

21D
Digit numberof frequency
range◆ 211 Voltage range
Sets the voltage range (primary voltage). Change of this setting also sets the measurement range ofactive power and reactive power automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：6600V (3φ3W，110V input), 110.0V (1φ3W), 3300V (1φ2W，110V input),
220.0V (3φ3W/1φ2W，220V input)
Setting factor
150V (110V)
300V (220V，220V/110V)
500V (380V/110V)
600V (440V/110V)
600V (460V/110V)
600V (480V/110V)
1200V (880V/110V)
1500V (1100V/110V)
2400V (1650V/110V)
3000V (2200V/110V)
4500V (3300V/110V)
9000V (6600V/110V)
15kV (11kV/110V)
18kV (13.2kV/110V)
18kV (13.8kV/110V)

24kV (16.5kV/110V)
25kV (18.4kV/110V)
30kV (22kV/110V)
45kV (33kV/110V)
90kV (66kV/110V)
120kV (77kV/110V)
150kV (110kV/110V)
180kV (132kV/110V)
210kV (154kV/110V)
270kV (187kV/110V)
300kV (220kV/110V)
400kV (275kV/110V)
500kV (380kV/110V)
750kV (550kV/110V)

New setting

Current setting
Setting No.
Voltage range

Setting factor

New setting

Setting No.

Current setting◆ 212 Digit number of voltage range
Sets the digit number of voltage range.
Selection by ＋ and － , set value is updated by SET .
Default setting：4 digits
The unit may be changed if the number of digits is changed.
Example) 6600V
6.60kV＋

4 digits＋－

3 digits－

Digit number of voltage range

42

SFLC-212-173◆ 213 Current range
Sets the current range (primary current). Change of this setting also sets the measurement range ofactive power and reactive power automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：100.0A (3φ3W), 500.0A (1φ3W), 50.00A (1φ2W)
Current-measurement range

5A
6A
7.5A
8A
10A
12A
15A
20A
25A
30A
40A
50A

60A
75A
80A
100A
120A
150A
200A
250A
300A
400A
500A
600A

750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A
2500A
3000A
4000A

5000A
6000A
7500A
8000A
9000A
10kA
12kA
15kA
20kA
30kA

◆ 214 Current display intrinsic sensitivity
Sets the full scale of current meter.
The setting range is 40 to 120％ of CT ratio. And, it can select from the following values.
Selection by ＋ and － , set value is updated by SET .
Setting factor
New setting
Default setting：100.0A (3φ3W), 500A (1φ3W), 50.0A (1φ2W)
The measurement range (current, active
power, reactive power) which can be
set. (×10n)
1.□ 1.0／1.2／1.4／1.5／1.6／1.8
2.□ 2.0／2.4／2.5／2.8
3.□ 3.0／3.2／3.6
4.□ 4.0／4.2／4.5／4.8
5.□ 5.0／5.6
6.□ 6.0／6.4
7.□ 7.2／7.5
8.□ 8.0／8.4
9.□ 9.0／9.6

Example)In case of CT ratio＝100.0A.
･40％ of 100A is 40A.
･120％ of 100A is 120A.
A measurement range can beselected within the limitsof 40 to 120A.
Therefore, a left table
A measurement range can beselected from
40/42/45/48/50/56/60/64/72/
75/80/84/90/96/100/120A.

◆ 215 Digit number of current range
Sets the digit number of current range.
Selection by ＋ and － , set value is updated by SET .
Default setting：4 digits
The unit may be changed if the number of digits is changed.
Example) 1000A
1.00kA
Current setting＋
4 digits＋－

Setting No.

Current setting

Current display intrinsic
sensitivity

New setting

Setting No.

3 digits－

Setting factor
Digit number of current range

43

SFLC-212-173◆ 216 Active power polarity
A swing display of active power meter can be selected from P (one-way deflection) and － (both deflection).
Selection by ＋ and － , set value is updated by SET .
Default setting：P (one-way deflection)
New setting＋

P：One-waydeflection＋－

Both swing display－：Bothdeflection

One-way deflection
display－<Note>
Setting No.
When the power polarity setting is changed, thescale of display and analog output will change.
Example) When the primary power value is 1200kW
Current setting
Power polarity
and the analog output is DC4mA to 20mA.
・ One sided swing setting: 0 to 1200kW / DC4mA to 20mA・ When both shakes are set: -1200kW to 0 to 1200kW / DC4mA to 12mA to 20mA
◆ 217 Active power range
Set the full scale of active power meter. The setting range is 30 to 120％ of rated power.
And it can select from "214 current display intrinsic sensitivity" tables.
Selection by ＋ and － , set value is updated by SET .
(1000MW or more becomes 4 digits display fixation.)<Caution> In case of 220V direct input. Calculates by VT ratio＝2.
Default setting：3φ3W, 110V input：1200kW ，3φ3W, 220V input：40.00kW ，1φ3W：100.0kW
1φ2W, 110V input：150.0kW ，1φ2W, 220V input：10.00kW◆ 218 Digit number of active power range
New setting
Sets the digit number of active power range.
Selection by ＋ and － , set value is updated by SET .
Default setting：4 digits
The unit may be changed if the number of digits is changed.
Example) 1200kW
1.20MW
Current setting＋

4 digits＋－

Setting No.

3 digits

Setting factor－

Digit number of active power range◆ 219 Reactive power range
Sets the full scale of reactive power meter.
The setting range is 30 to 120％ of rated reactive power.
And it can select from "214 current display intrinsic sensitivity"
tables. Selection by ＋ and － , set value is updated by SET .
(1000Mvar or more becomes 4 digits display fixation.)

Setting No.

New setting<Caution> In case of 220V direct input. Calculates by VT ratio＝2.
Default setting：3φ3W, 110V input：600.0kvar
3φ3W, 220V input：20.00kvar
1φ3W：50.00kvar
1φ2W, 110V input：75.00kvar
1φ2W, 220V input：5000var

Setting factor
Current setting
Reactive power range

44

SFLC-212-173◆ 21A Digit number of reactive power range
New setting
Sets the digit number of reactive power range.
Selection by ＋ and － , set value is updated by SET .
Default setting：4 digits
The unit may be changed if the number of digits is changed.
Example) 1000kvar
1.00Mvar
Current setting＋＋

4 digits－

Setting No.

3 digits

Setting factor－

Digit number of active power range◆ 21B Power-factor range
A power-factor measurement range can be selected from 0.5 to 1 to 0.5/0 to 1 to 0.
Change of this setting also sets the analog output range of a power-factor automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：0.5 to 1 to 0.5＋＋

0.5 to 1 to 0.5－

0 to 1 to 0－

◆ 21C Frequency range
A frequency measurement range can be selected from 45 to 55Hz/55 to 65Hz/
45 to 65Hz. Change of this setting also sets the analog output range
of a frequency automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：45 to 65Hz
50：
45 to 55Hz＋60：
55 to 65Hz－＋－

Setting factor

New setting

55：
45 to 65Hz－＋
Setting No.
Current setting
Frequency range◆ 21D Digit number of frequency range
Sets the digit number of frequency range.
Selection by ＋ and － , set value is updated by SET .
Default setting：3 digits
New setting＋

4 digits＋－

Current setting
3 digits
Setting No.

－

Setting factor
Digit number of frequency range

45

SFLC-212-173(2) 221A to 227A Analog output setting 【With option】
Various setting of analog output is performed.
222A

221A
MAX/MIN

Ao1 output
factorRESET/SHIFTRESET/SHIFT
MAX/MIN

223A

MAX/MIN

Ao2 output
factor227A
Low input cut

MAX/MINRESET/SHIFT

Ao3 output
factorRESET/SHIFTRESET/SHIFT
MAX/MIN

226A
Reactive power
output intrinsic
sensitivity

RESET/SHIFT
MAX/MIN◆ 221A to 223A AO (analog output) 1 to 3 output factor.
Sets the output factor of each analog outputs.
Selection by ＋ and － , set value is updated by SET .

224ACurrent output
intrinsic
sensitivity
225A
Active power
output intrinsic
sensitivity

RESET/SHIFT
MAX/MIN

New setting

Default setting：Current setting

AO1 A(V) (3φ3W)，A(U) (1φ3W U-W-N)，A (1φ2W)
AO2 V(UV)(3φ3W)，V(UN)(1φ3W U-W-N)，V (1φ2W)
AO3 W (3φ3W / 1φ3W / 1φ2W)

Setting No.
Ao1 output factor

OFF：
Nothing－＋

9：DA(W)

10：MDA(U)

＋
－
－
＋
＋
－
－
＋

1：V(UV)

8：DA(V)

11：MDA(V)

18：Hz

＋
－
－
＋
＋
－
－
＋

2：V(VW)

7：DA(U)

12：MDA(W)

17：cosφ◆ 224A Current output intrinsic sensitivity,
225A Active power output intrinsic sensitivity,
226A Reactive power output intrinsic sensitivity.

＋
－
－
＋
＋
－
－
＋

3：V(WU)

6：A(W)

13：W

16：var

＋
－
－
＋
＋
－
－
＋

＋

4：A(U)－

5：A(V)＋

14：DW－

15：MDWAt 3φ3W

New setting

Current setting

Setting No.

Output intrinsic sensitivity (％ of rated input power value toan output upper limit value) is set about each analog output ofcurrent, active power, and reactive power.
The setting range can be selected from the following.
Current：40.0 to 120.0％ (0.1％ step)
Active power ：30.0 to 120.0％ (0.1％ step)
Reactive power：30.0 to 120.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：100.0％(Current, Active power)，50.0％(Reactive power)

Power-output intrinsic
sensitivity,

Example) 3φ3W, 9000V, 100.0A, 1200kW range, Analog output DC4 to 20mA･Set 100.0％. Analog output is DC20mA in 1200kW of primary inputs.
･Set 80.0％. Analog output is DC20mA in 1200kW×0.8＝960kW of primary input.
◆ 227A Low input cut
It is the function which makes analog output a lower
limit at the case of a minute input (input whichcorresponds to 0.5％ or less).
A function can be selected from ON (Use) and OFF(Non-use).
Selection by ＋ and － , set value is updated
by SET .
Default setting：OFF (Non-use)

New setting
Current setting
Setting No.
Setting factor

Low input cut
46

SFLC-212-173(3) 231P to 232P Pulse output setting 【With option】
Various setting of a pulse output is performed.
231P
Pulse output factor

MAX/MIN

232P
Pulse unit

RESET/SHIFT

MAX/MINRESET/SHIFT

New setting
Setting No.
Current setting◆ 231P Pulse output factor,
Sets the output factor of pulse output.
Selection by ＋ and － , set value is updated by SET .
Default setting：Wh
OFF：
Nothing

＋

＋

1：Wh

＋

2：-Wh

－

－

－

－

－

－

3：varh
(LAG)
－ ＋

＋

6：-varh(LEAD)＋

5：-varh(LAG)

4：varh(LEAD)＋

Pulse output factor◆ 232P Unit of pulse output
New setting
Sets the pulse unit of pulse output.
A pulse unit can be selected from four types.
Current setting
The pulse unit that can be selected is decided withfull-load power.
Setting No.
Default setting：10kWh/pulse [3φ3W, 110V input]
1kWh/pulse [1φ3W/1φ2W, 110V input]
0.1kWh/pulse [3φ3W/1φ2W, 220V input]

Pulse unit
(4) 241 External operation input setting【With option】
Various setting of external operation input is performed.
◆ 241 External operation input function
The function of each external operation
input (alarm reset, maximum / minimum reset,
all reset) can be selected.
Selection by ＋ and － , set value is

Setting factor
Setting No.

New setting

updated by SET .
Default setting (With alarm-output option)
Alarm reset
Alarm reset
－
＋

＋
－

Maximum and minimum
are displayed byturns.
Flashing of RESET.
(Maximum / minimum
reset)

Maximum /minimum reset

All reset(Alarm,Maximum/minimum)

ALARM and RESET aredisplayed by turns.
(Alarm reset)－＋

Current setting

・About the setting display in an external operation input function
Function
Alarm reset
Maximum /minimum reset
All reset

The contents of a display at the function setting"ALARM" and "RESET" are displayed by turns by 7 segment
display.
Flashing of RESET, by 7 segment display.
And, maximum and minimum of guidance displayed by turns.
"ALL" and "RESET" are displayed by turns by 7 segment
display.

47"Current setting" "New setting"display point
display point

Sub monitor(right)Main monitor

SFLC-212-173(5) 251 to 25C Measurement ON/OFF setting
Measurement display ON/OFF setting of each measurement factor is performed.
Selection by ＋ and － , set value is updated by SET .
Default setting：ON (All measurement factors)
251

MAX/MIN

Voltage

RESET/SHIFTRESET/SHIFT
MAX/MIN

256
Reactive
power
257
Power factor

25C
Power
transmission
var-hour

MAX/MIN

252

RESET/SHIFT

Current

RESET/SHIFT
MAX/MIN

255
Demand power

258

MAX/MINRESET/SHIFT

Frequency

RESET/SHIFT
MAX/MIN

25B
Power
receivingvar-hourNew setting
Setting No.
Current setting

253
Demandcurrent

MAX/MINRESET/SHIFTRESET/
SHIFT
MAX/
MIN

RESET/SHIFT
MAX/MIN

Active power

MAX/MINRESET/SHIFT

259
Powerreceiving
watt-hourRESET/SHIFT
MAX/MIN

254

25A
Power
transmission
watt-hourRESET/
SHIFT
MAX/
MIN

Voltage ON/OFF(6) 261 to 262 Initialization of watt-hour and setting value.
All reset of watt-hour, and initializes the each settings (it returns to a default setting).
261
Setting initialization

MAX/MINRESET/SHIFT

262
Watt-hour reset

MAX/MINRESET/
SHIFT◆ 261 Initialization of setting value
Initialize the settings of setting 1 and setting 2 (return to the default settings).
Pushing SET for 3 seconds or longer to initialize the settings of setting 1 and setting 2.
(The set value of setting 3 does not return to the default value)SET
3 seconds

Setting No.

Setting No.

Completion of
initializing(Before initialization)(Initialization completion)

Initialization of setting value◆ 262 Watt-hour reset
Integrated value of each electric-energy display is cleared (＝0). By pushing SET for 3 seconds,
all integrated value (Wh, -Wh, var(LAG), -var(LAG), var(LEAD), -var(LEAD)) is all cleared.
Wh and varh aredisplayed byturns.

Wh and varh aredisplayed byturns.

SET 3 seconds
Setting No.
After clear
completion
Clear completion

Setting No.
(Before clear)

Watt-hour reset48(Clear completion)

SFLC-212-173

5.3.3

Setting mode 3
SET and DISPLAY
3 seconds
DISPLAY

311, 312
Input circuit setting
MODE

Display mode

DISPLAY

321
Tidal current measurement setting
MODE

DISPLAY

331 to 336
Analog output adjustment
MODE

Setting mode 3

Setting mode 3 is selected by pressing SET and DISPLAY switches continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, a display and output of measurement are not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

49

SFLC-212-173(1) 311 to 312 Input circuit setting
Sets the input circuit and phase wire and input voltage / phase-voltage full scale.
311
Input circuit phasewire change

312

MAX/MIN

MAX/MIN

Input voltage

RESET/SHIFTRESET/
SHIFT◆ 311 Input circuit phase wire change
Sets the input circuit and phase wire (1φ3W).
Selection by ＋ and － , set value is updated by pushing SET
3 seconds or more.
Default setting：3φ3W (No designation)

New setting
Current setting
Setting No.

<Caution>
・When this setting is changed, it will become the default settingof the input circuit after all set values changing.
・The right measurement cannot be performed if setting of actual
connection and phase wire are different.
＋

3φ3W

－

1φ3W
(U-N-W)＋－

1φ3W(U-N-V)＋

1φ3W(V-N-W)

－

＋

1φ2W

－

Input circuit andphase wire change－＋◆ 312 Input voltage
Set the input voltage (3φ3W, 1φ2W) or phase-voltage full scale (1φ3W).
In 3φ3W/1φ2W and 1φ3W, the contents of a setting are different.
Selection by ＋ and － , set value is updated by SET .
Default setting：150V (1φ3W)
Default setting：110V (3φ3W，1φ2W or no designation)
・3φ3W，1φ2W

・1φ3W
＋

110V

＋

＋

150V

220V

＋
300V

－

－

－

－

Setting of 300V, Output is AC0 to 150V/DC4 to 12mA.
Setting of 150V, Output is AC0 to 150V/DC4 to 20mA.
New setting
Setting No.
Current settingUN
Setting No.

New setting

WN alternate display(Phase display U-N-W)

Current setting

Input-voltage rating (3φ3W)

Phase-voltage full scale (1φ3W)

50

SFLC-212-173(2) 321 Tidal current measurement setting
Sets the with or without of tidal current measurement.
◆ 321 Tidal current measurement
By measurement of reactive power and a power-factor,
New setting
0 (general measurement) or 1 (tidal current measurement
which was conscious of power transmission/power
Current setting
receiving) can be selected.
Selection by ＋ and － , set value is updated by SET .
Setting No.
Default setting：0 (General measurement)＋＋

0：Generalmeasurement

1：Tidal current

measurement－－

Tidal current measurement(General measurement)
Display

Analog output[mA] LEAD 0 to LAG 0/4 to 20mA

LEAD 0 to LAG 0/LEAD 0 to LAG 0

LAG 0

20

cosφ=1

12

LEAD 04

Input
φ=-90° φ=0° φ=90°
φ=180°
φ=270°[cosφ]
LAG
cosφ LEAD
LEAD cosφ
1
0
1
0
0

Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]
LAG
cosφ LEAD
LEAD cosφ
1
0
1
0
0

Power receiving

Power receiving

Power transmission

Power transmission(Tidal current measurement)
Display

LEAD 0 to LAG 0/LEAD 0 to LAG 0

Analog output[mA] LEAD 0 to LAG 0/4 to 20mA

LAG 0

20

cos180°=1
LEAD
0
LAG

12

cos0°=1
LEAD 04
Input
φ=-90° φ=0°φ=90°φ=180°
φ=270°[cosφ]LEAD cosφ LAG,LEAD cosφ LAG
1
0
0
1
0
0

Input
φ=-90° φ=0°φ=90°φ=180°
φ=270°[cosφ]LEAD cosφ LAG,LEAD cosφ LAG
1
0
0
1
0
0

Power receiving

Power receiving

Power transmission

51

Power transmission

SFLC-212-173(3) 331 to 336 Analog output adjustment 【With option】
The bias and span of each analog output are adjusted.
331
Ao1 BIAS
adjustment

MAX/MINRESET/SHIFT

336
Ao3 SPAN
adjustment

MAX/MINRESET/SHIFTRESET/SHIFT
MAX/MIN

332
Ao1 SPAN
adjustment

335
Ao3 BIAS
adjustment

MAX/MINRESET/SHIFTRESET/SHIFT
MAX/MIN

333
Ao2 BIAS
adjustment
RESET/SHIFT
MAX/MIN

334
Ao2 SPAN
adjustment◆ 331 AO (Analog output) 1 bias adjustment,
333 AO (Analog output) 2 bias adjustment,
335 AO (Analog output) 3 bias adjustment.
The bias of each analog output is adjusted.
Setting range：±10.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
◆ 332 AO (Analog output) 1 span adjustment,
334 AO (Analog output) 2 span adjustment,
336 AO (Analog output) 3 span adjustment.
The span of each analog output is adjusted.
Setting range：±10.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .

52

Ao1 span adjustment

SFLC-212-173

6. Specification
6.1 Specification and intrinsic error.
Input circuit
Input
3-phase 3-wire
AC110V，220V common use. AC5A 50/60Hz
Single-phase 2-wireSingle-phase 3-wire
AC100-200V (21), AC5A 50/60Hz21Note ( ) The rated voltage of each phase and N phase is 100V. However, the full scale of a bar graph is 300V.
Intrinsic error (22)
Measurement
item

Measurement range /Display specification

Digitaldisplay

Voltage

AC150V to 750kV±0.5％

Current

Maximum demand, Demand, Instant
AC5A to 30kA

±0.5％

Active power

Reactive
power

Maximum demand, Demand, Instant
150.0W to 1200MW (Range select)
In voltage and current range.
One-way deflection or bothdeflection can be setting.
LEAD,LAG 150.0var to 1200Mvar(Range select)
In voltage and current range.

Maximum Minimum
measur- measurNotes
Analog output
ement
ement
(23)
27
27
（ ） （ ）
±0.5％○○
UV-VW-WU line change (24)
U-V-W phase change (25)
Apart from a measurement
±0.5％
○
○
range, range setting of adisplay and an output ispossible.

±0.5％

±0.5％

○

○

±0.5％

±0.5％

○

○

Power factor

LEAD 0.500 to 1.000 to LAG 0.500 or
LEAD 0.000 to 1.000 to LAG 0.000
Range select

±2.0％

±2.0％

○

○

Frequency

45.0 to 55.0Hz or
55.0 to 65.0Hz or
45.0 to 65.0Hz
Range select

±0.5％

±0.5％

○

○

Range of an analog output
can be set as indication
independently.
(26)
Range of an analog output
can be set as indication
independently.
In case input is below
20％ of voltage range or
below 2％ of current
range：cosφ=1. (Output
is cosφ=1 equivalence.)
0.0Hz in case input isbelow 20％ of voltage
range. Output is alower limit value.
(Lower limit value -1％：％ for output span)

Intrinsic error (22)
Measurement
item

Watt-hour

var-hourMeasurement range /Display specification
Display：Integer, 5 digit.
Multiplier：Integral number time of 10.
Expansion display is possible to the
3rd place below a decimal point.
Electric power is integrated. (Power
receiving，Power transmission)
Display：Integer, 5 digit.
Multiplier：Integral number time of 10.
Expansion display is possible to the
3rd place below a decimal point.
Integrating reactive power of powerreceiving. (LAG･LEAD)
Integrating reactive power of powertransmission. (LAG･LEAD)

Digitaldisplay

Maximum Minimum
measur- measurPulse output
ement
ement
(23)
（27） （27）Power factor
1：±2.0％Power factor
1：±2.0％Power factor
0.5：±2.5％

Power factor
0.5：±2.5％

Power factor
0：±2.5％

Power factor
0：±2.5％

Power factor
0.87：±2.5％

Power factor
0.87：±2.5％

Notes
Conformity with normal
watt-hour meter.
Setting range of pulse
output unit (kWh/pulse)
is referred tooption-specification.

Setting range of pulse
output unit (kvarh/pulse)
is referred to optionspecification.

Note (22) If this unit directly measures an inverter output of cycle control, SCR phase angle control or PWM control,
an error may increase due to its operation principle.
Note (23) Analog output, pulse output, alarm output and external operation input are options.
Note (24) Single-phase 3-wire (U-N-W)：UN-WN-UW, Single-phase 3-wire (U-N-V)：UN-VN-UV, Single-phase 3-wire (V-N-W)：VN-WN-VW, Single-phase 2-wire：With no phase display.
Note (25) Single-phase 3-wire (U-N-W)：U-N-W, Single-phase 3-wire (U-N-V)：U-N-V, Single-phase 3-wire (V-N-W)：
V-W-N, Single-phase 2-wire：With no phase display.
Note (26) At the case of one-way deflection setting of bar graph. Digital meter measures reverse power to -15%
full scale.
Note (27) It can usually check the maximum value and the minimum value by MAX/MIN switch operation from a display.

53

SFLC-212-173

Item
Bar graphdisplay

OperatingmethodIntervalsetting

The factor
in whichdisplaysetting ispossible

OptionSpecification
Bar graph display of the main-monitor factor is done. (Watt-hour and var-hour exclude)
A display of a sub monitor factor can also be set.
Current, Voltage ：Effective value computing type.
Demand ammeter：Arithmetic method according with bimetallic type.
Demand power meter：Arithmetic method according with bimetallic type, or average value within the
demand time limit. (One side is selected by setting.)
Active power, Reactive power, Watt-hour, var-hour：Time-division multiplication method.
Power factor：Calculates for active power and reactive power.
Frequency：Zero cross cycle computing type.
Demand0 second / 5 seconds / 10 seconds / 20 seconds / 30 seconds / 40 seconds /current
50 seconds / 1 minute / 2 minutes / 3 minutes / 4 minutes / 5 minutes /6 minutes / 7 minutes / 8 minutes / 9 minutes / 10 minutes / 15 minutes /Demand power
20 minutes / 25 minutes / 30 minutes (95％ time limit)
Voltage (Each phase and line), Current (Each phase), Demand current (Each phase),
Maximum demand current (Each phase)， Active power, Demand power,
Main monitor Maximum demand power, Reactive power, Power factor, Frequency,
Watt-hour (Power receiving, Power transmission),var-hour (Power receiving LAG/LEAD, Power transmission LAG/LEAD)
Sub monitor Voltage (Each phase and line), Current (Each phase)， Active power， Reactive power，(Left)
Power factor
Voltage (Each phase and line), Current (Each phase)， Demand current (Each phase)，
Sub monitor
Maximum demand current (Each phase)， Active power， Demand power，Maximum demand power，(Center)
Reactive power， var-hour (Power receiving LAG/LEAD, Power transmission LAG/LEAD)
Voltage (Each phase and line), Current (Each phase)， Demand current (Each phase)，
Sub monitor
Maximum demand current (Each phase)， Active power，Demand power， Maximum demand power，(Right)Power factor，Frequency， Watt-hour (Power receiving, Power transmission)
Voltage (Each phase and line), Current (Each phase)， Demand current (Each phase)，
Bar graph
Maximum demand current (Each phase)，Active power，Demand power， Maximum demand power，Reactive power， Power factor， Frequency
Analog output (3 sets). Pulse output. Alarm output. External operation change input.

● About power and reactive power full-scale range selection.
The power range and reactive power range is automatically decided in a current range and voltage range.
The full scale range of a bar graph can be selected out of the following range within a 30 to 120％, assumingthat the rated power (VT ratio × CT ratio)(28) is 100％.
1.0 / 1.2 /1.4 / 1.5 / 1.6 / 1.8 / 2.0 / 2.4 / 2.5 / 2.8 / 3.0 / 3.2 / 3.6 / 4.0 / 4.2 / 4.5 / 4.8 / 5.0 /5.6 / 6.0 / 6.4 / 7.2 / 7.5 / 8.0 / 8.4 / 9.0 / 9.6 ×10n
Example) (VT ratio)×(CT ratio)＝1200kW
A full scale range can be selected from the following.
480 / 500 / 560 / 600 / 640 / 720 / 750 / 800 / 840 / 900 / 960 / 1000 / 1200
Note (28) Assume VT ratio is “2” for calculation in case of 220V input specifications.
● Measurement is possible range.
Measurement
factorVoltage
Current
Demand current
Active power
Demand power

Input (29)AC0 to 150V [AC0 to 300V]
AC0 to 5A±1kW [±2kW]

Measurement is possible range
Display
Analog output
101％ of meter full scale
101％ of output span
120％ of meter full scale (30) 120％ of output span
200％ of meter full scale (30) 120％ of output span
120％ of meter full scale (30) -1％ and 120％ of output span
200％ of meter full scale (30) -1％ and 120％ of output span

LEAD 1 to 0 to LAG 1kvar
120％ of meter full scale (30) -1％ and 120％ of output span
[LEAD 2 to 0 to LAG 2kvar]
LEAD 0 to 1 to LAG 0
LEAD 0.000 to 1 to LAG 0.000
0％ and 100％ of output span
Power factor
LEAD 0.5 to 1 to LAG 0.5
LEAD 0.490 to 1 to LAG 0.490-1％ and 101％ of output span
45 to 55Hz
44.9 to 55.1Hz
Frequency
55 to 65Hz
54.9 to 65.1Hz-1％ and 101％ of output span
45 to 65Hz
44.8 to 65.2Hz
Note (29) [ ] is the 300V input case.
Note (30) If the number of display digits is exceeded in spite of the measurable range, it becomes to 9999
(four-digit display) or 999 (triple digit display).
Reactive power

54

SFLC-212-173

6.2

Performance.

Item
Accuracy
Accuracy of bar graph
Influence by temperatureCompliance standard

Safety

Display updating time
Display device
Display composition

LCD view angle

Backlight
Auxiliary supply
Rush current(Time constant)
Input consumption VA

Overload capacity

Insulation resistance
Voltage test
(Commercial frequency
withstand voltage)
JIS C 1102-1
JIS C 1111
JIS C 1216-1
JIS C 1263-1

Specification
Reference to measure specification and accuracy±10％ （％ for span）
Within accuracy by 23±10℃.
JIS C 1102-1：2007 ，JIS C 1102-2,-3,-4,-5,-7：1997
JIS C 1111：2006 ，JIS C 1216-1：2009 ，JIS C 1263-1：2009
JIS C 1010-1：2005
CATⅢ (The category to the measurement performed with fabric equipment)
Maximum circuit voltage 300V
Pollution degree 2 (Usually, environment which only contamination of non-conductivity
generates. However, temporary conductivity which originates indew condensation depending on the case occurs.)
About 1 second (Bar graph：0.25 seconds)
Main monitor
5 digit, character height 11mm
LCD
Sub monitor (Left)
4 digit, character height 6mm
Number, Character,
Sub monitor (Center),(Right) 5 digit, character height 6mm
Segment color：BlackBar graph
20 dots
For upper installation Upper view angle 10°，Lower view angle 60°，
(For lower view)
Right and left view angle 60°
For lower installation Upper view angle 60°，Lower view angle 10°，
(For upper view)
Right and left view angle 60°
Upper view angle and lower view angle 75°，
Wide viewing angle
Right and left view angle 75°
LED backlight：White (31)
Always-on, Auto off (after 5 minutes without operating), Always-off. Setting is possible.
White backlight can select brightness from five steps of 1 to 5. (Default：3)
AC85 to 264V 50/60Hz 10VA (Rated voltage, AC100/110V，200/220V)
DC80 to 143V
6W (Rated voltage, DC100/110V) for both AC and DC uses
Rated voltage AC110V 2.2A or less (About 2.5ms)
Rated voltage AC220V 4.4A or less (About 2.5ms)
Rated voltage DC110V 1.6A or less (About 2.5ms)
Voltage circuit
0.25VA or less (110V) ，0.5VA or less (220V)
Current circuit
0.1VA or less (5A)
Voltage circuit
2 times 10 seconds, 1.2 times continuation of rated voltage.
40 times 1 second, 20 times 4 seconds, 10 times 16 seconds,
Current circuit
1.2 times continuation of rated current.
1.5 times 10 seconds, 1.2 times continuation of rated voltage.
Auxiliary supply In case of DC110V, 1.5 times 10 seconds, 1.3 times continuation of
rated voltage.
Between electric circuits and case (Earth).
Between input and output and auxiliary supply.
Above 50MΩ at DC500V
Between analog output and pulse output and alarm output.
Between analog outputs are not insulation of minus common.
Between electric circuits and case (Earth).
AC2210V (50/60Hz) 5 seconds
Between input and output and auxiliary supply.
Between electric circuits and case (Earth).
Between input and output and auxiliary supply.

AC2000V (50/60Hz) 1 minuteBetween analog output and pulse output and alarm output.
Between analog outputs are not insulation of minus common.

31

Note( ) About white backlight.
The white backlight of this product is using white LED which combined the special phosphor and blue
LED. In the characteristics of this LED, color tone may be different for each product.

55

SFLC-212-173

Item
Impulse voltage test
(Lightning impulse
withstand voltage)
JIS C 1111
JIS C 1216-1
JIS C 1263-1

Noise-capacity
JEA B-402

Vibration
JIS C 1102-1
Shock
JIS C 1102-1
Construction
Material
Color
MassBlackout guarantee
Operating temperatureand humidity limits
Storage temperature
limitsInstallation altitude

Specification
Between electric circuits and case (Earth).
(An analog output is excluded)

6kV 1.2/50μs Both positive andnegative polarities, for 3 time each.

Between input and auxiliary supply.
(Grounds an output.)

5kV 1.2/50μs Both positive andnegative polarities, for 3 time each.

(1) Oscillatory surge voltage
When a damping vibration waveform (peak voltage 2.5kV, frequency 1MHz±10%) isrepeatedly applied, the measurement error should be within 10% and no malfunction
should occur.
Voltage input circuit (Normal / Common), Current input circuit (Common),Auxiliary supply circuit (Normal / Common)(2) Square wave impulse noise
If a noise (1μs, 100ns width) is repeated and added, a measurement error should bewithin 10％. And, there needs to be no malfunction.
Auxiliary supply circuit (Normal / Common) Over 1500V
Voltage input circuit (Normal / Common) Over 1500V
Current input circuit (Common) Over 1500V
Pulse output (Common) Over 1000V
Alarm output (Common) Over 1000V
Operation input (Common) Over 1000V
Analog output circuit (Induction) Over 1000V(3) Electric wave noise
If intermittence irradiation of the electric wave of a 150MHz,400MHz band is doneby (5W, 1m), a measurement error should be within 10%. And, there needs to be no
malfunction.
(4) Electrostatic noise
Measurement error should be within 10% at contact discharge 8kV and air discharge
15kV, and malfunction shall not occur.
Sweep vibration frequency range：10 to 55 to 10Hz， Displacement amplitude：0.15mm，
Number of sweep：5， Sweep velocity：1 octave /minutePeak acceleration：490m/s2 ，Waveform of pulse：Sine half wave，duration of pulse：11ms
Number of shock：It is each 3 times about a forward reverse to 3 shaft orientations(right-angled to mutual). (Total 18 times)
Dimension：110mm(Width)×110mm(Height)×103.5mm(Depth) Body diameter：99mmφ
With terminal cover, Protection code IP40
Case，Cover：ABS(V-0) ，Terminal block：PBT ，Terminal cover：Polycarbonate
Black (Munsell N1.5)
Approx. 600g
Maximum value, Minimum value, Integrates value, Each setting value.
Data hold by nonvolatile memory.
-10 to +55℃, 30 to 85％ RH, Non condensing.
-25 to +70℃
The altitude of 2000m or less.

56

SFLC-212-173

6.3 Option

Item

Specification
3 circuits (Minus common)
DC4 to 20mA (Below 550Ω)
Voltage (Each phase and line)， Current (Each phase)，
Demand current (each phase)， Maximum demand current (Each phase),
Output factor
Active power， Demand power, Maximum demand power， Reactive power，Power factor， Frequency
Response time
1 second or less (Time within ±1％ of final constant value.)
Output ripple
Within the double precision of accuracy (％ for output span)
Possible output ：Watt-hour or var-hour.
Output form
：Optical MOS-FET relay. 1a contact
Contact capacity：AC，DC125V, 70mA (Resistance load, inductive load)
Pulse width：250±10ms (Output pulse width when the output pulse period of rated active power
constitutes speed more than 2 pulse/second by setting of an voltage measurement
range, a current-measurement range, and an output pulse unit is set to 100 to
130ms.)
Setting of output pulse unit is possible by the next range.
■ 3-phase 3-wire：Full load power (kW,kvar)＝√3×Rated voltage(V)×Rated current(A)×10-3■ Single-phase 3-wire：Full load power (kW,kvar)＝2×Rated voltage(V)×Rated current(A)×10-3■ Single-phase 2-wire：Full load power (kW,kvar)＝Rated voltage(V)×Rated current(A)×10-3
Full load power (kW,kvar)
Output pulse unit. kWh(kvarh)/pulse
Multiplying factor
Below 1
0.1
0.01
0.001
0.00010.01 (32)Over 1
Below 10
1
0.1
0.01
0.001
0.1
Over 10
Below 100
10
1
0.1
0.01
1
Over 100
Below 1,000
100
10
1
0.1
10
Over 1,000
Below 10,000
1,000
100
10
1
100
Over 10,000 Below 100,000
10,000
1,000
100
10
1,000
Over 100,000 Below 1,000,000 100,000
10,000
1,000
100
10,000Alarm factor：Demand current, Demand power, Voltage, Alarm OFF. Possible to setting one of them.
Reset method：Automatic reset or Manual reset (Setting)
Output contact ：No-voltage a contact (OR of each phase detection)
Contact capacity：AC250V 5A, DC125V 0.3A (Resistance load) AC250V 2A, DC125V 0.1A (Inductive load)
Alarm factor
Item
Specification
Demand measurement value ≧ Upper limit setting value,
Function
Alarm display, Alarm output.
Demand current,
Setting accuracy ±0.5％ (％ for full scale )Demand power
Setting range
5 to 100％ to the maximum scale. (1％ step)
Measurement value ≧ Upper limit setting value,
Alarm display, Alarm output.
Function
Measurement value ≦ Lower limit setting value,
Voltage
Alarm display, Alarm output.
Setting accuracy ±0.5％ (％ for full scale)Setting range
Using a full scale as 150％. 30 to 150％ (1％ step)Three types of following functions can be operated by adding a
Function
voltage signal from the outside in addition to switch operation.
Alarm output is reset (output OFF).
Alarm reset
Please refer to "4.3.6 Reset" about operation by the switch.
The maximum/minimum value is reset (it updates to theMaximum / Minimum value
instantaneous value at the time).
reset
Please refer to "4.3.6 Reset" about operation by the switch.
Resets all of the alarm output and maximum/minimum value.
All reset
Please refer to "4.3.6 Reset" about operation by the switch.
Minimum operation pulse width
300ms ，Continuation applying is possible.
Input rating becomes the same as that of auxiliary supply.
AC100/110V 0.4VA，AC200/220V 1.4VA，DC100/110V 0.4W
Rated input
AC DC two ways.
Contact capacity：About 3mA (AC,DC100/110V)，
About 6mA (AC200/220V)
Number of output
Output specification

Analog output

Pulse output

Alarm output

External
operation
input

Note(32) Although multiplying factor is 0.01, a multiplying factor display will be 0.1.
(The place by the integer is 4 digits display. An enlarged display is 4 digits below decimal point.)
57

SFLC-212-173● Caution on the use of external display selection input (option)
External power consumption is 0.4VA at AC110V or 1.4VA at AC220V or 0.4W at DC110V.
In case a relay or a switch is used for power-supply supply, please use the thingof about 1mA of the minimum application loads.

P N
28
29

7. Maintenance and check7.1 Trouble shooting

Symptoms
Indicator does not
display.
Backlight does notlights.

Have a margin of
measurement error.

Analog output is not
outputted.
Pulse output is not
outputted.
Alarm output does not
return.
Settings changed.

Possible causes
The power supply is not supplied.
(Not connected. or voltage is low)
Measurement display ON/OFF setting is set to OFF.
Trouble of device.

Remedial measures
Check the auxiliary supply.
Again, a power supply is supplied.
Check the setting.
Replace the device.

The setting is set to auto off or always off.

Check the setting.

Setting of a range is not right.
Setting of a input voltage is not right.
Wiring is not right.
Outside the rated frequency (45 to 65Hz).
Cycle control, SCR phase angle control, PWM control,
or other inverter output is measured.
Analog output is set to OFF or measurement factor isset to OFF.
Pulse output is set to OFF or measurement factor isset to OFF.

Please set again.
Check the setting.
Check the wiring.
Cannot be used.
Cannot be used.
Check the setting.
Check the setting.

The return method is a "manual reset".

Check the setting.

It changed an input circuit or input voltage setting.

Please set again.

58

SFLC-212-173

7.2 Test
SET and MODE
3 seconds
DISPLAY

Input wiring checkMODE

DISPLAY

Test mode is selected by pressing SET and MODE switches
continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode bypressing DISPLAY switch.

Alarm output check
MODE

Displaymode

DISPLAY

Analog output checkMODE

DISPLAY

Pulse output checkMODE

DISPLAYLCD display check
MODE(1) Input wiring check
It becomes a wiring check screen and can check the connection status of a voltage input and a current input.

Flicker

Positive phase sequence display.

Negative phase sequence display.

The example of a display (3φ3W)
Main monitor(33)：Positive phase sequence,“ ”(Positive),
Negative phase sequence, “ ”(Negative),
With no input, “－－－－”Sub monitor (Left) (34)：Power of U phase.
Sub monitor (Right) (34)：Power of W phase.
Note(33) It is set to “－－－－” if it is used in 1φ3W and 1φ2W circuit.
Note(34) The phases displayed by setting of an input circuit phase wire change are different at 1φ3W.
There is no phase display at 1φ2W.

59

SFLC-212-173(2) Alarm output check 【 With an option 】
Even if this product does not have input, it can perform ON/OFF test of an alarm output (relay-contact output).
Whenever it pushes SET , ON and OFF change.
Default：OFF

SET

The status of lights
at the output time.

Output OFF

Output ON
Alarm test(3) Analog output check 【With an option】
Even if this product does not have input, it can test analog output (three circuits).
Ao1 check

MAX/MINRESET/SHIFT

Ao2 check

MAX/MINRESET/SHIFT◆ AO (analog output) 1 to 3 check.
Sets the output factor about each analog output.
It selects 0％(4mA), 50％(12mA), and 100％(20mA)
with ＋ or － switch. If SET is pushed,analog output will output.
Default：0％ (4mA)
Analog output
factorAo3 check

MAX/MINRESET/
SHIFT

Output value

The status oflights at theoutput time.

(4) Pulse output check 【With an option】
Even if this product does not have input, it can test pulse output.
A push on SET outputs the pulse of a rated power. Pulse unit are the settings of "232P Pulse unit".
Default：OFF

SET
The status of lights
at the output time.

60

SFLC-212-173(5) LCD display check
It can check a liquid crystal display.
Default：Main monitor “LCD” display

Whenever it pushes SET , a display changes.

SET

61

SFLC-212-173
Appendix table 1-1
Active power range, watt-hour multiplier rate table (Three-phase three-wire)
180kV
150kV
45kV
30kV
300kV
255kV
210kV
750kV
500kV
105kV
90kV
375kV
V range(VT550kV/110V) (VT380kV/110V) (VT275kV/110V) (VT220kV/110V) (VT187kV/110V) (VT154kV/110V) (VT132kV/110V) (VT110kV/110V) (VT77kV/110V) (VT66kV/110V) (VT33kV/110V) (VT22kV/110V) Multiplying
Multiplying
factorfactor
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
A range
5A

5000 k

6A

6000 k

7.5A
8A
×100

10A

10.00 M
12.00 M

15A

15.00 M

20A

20.00 M

25A

25.00 M

30A

30.00 M

2000 k2400 k4000 k(3750)
4000 k

3000 k5000 k4000 k6000 k

4800 k7500 k6000 k

10.00 M

8000 k

14.00 M
(12.50)15.00 M

10.00 M

3200 k12.00 M

20.00 M

16.00 M
20.00 M

30.00 M

24.00 M

40.00 M(37.50)40.00 M

30.00 M

50.00 M

40.00 M

60.00 M

48.00 M

75.00 M

60.00 M

100.0 M

80.00 M

140.0 M
(125.0)150.0 M

100.0 M

200.0 M

160.0 M

250.0 M

200.0 M

300.0 M

240.0 M

400.0 M
(375.0)400.0 M

300.0 M

450.0 M

360.0 M

500.0 M

400.0 M

600.0 M

480.0 M

1500A

750.0 M

600.0 M

1600A

800.0 M

640.0 M

900.0 M

720.0 M

1000 M

800.0 M

1800 k(1700)
2400 k(2040)
2800 k(2550)
2800 k(2720)
3600 k(3400)
4200 k(4080)
5600 k(5100)
7200 k(6800)
9000 k
8500
12.00 M
(10.20)14.00 M
(13.60)18.00 M
(17.00)24.00 M
(20.40)28.00 M
(25.50)28.00 M
(27.20)36.00 M
(34.00)42.00 M
(40.80)56.00 M
(51.00)72.00 M
(68.00)90.00 M
(85.0)
12.0 M(102.0)140.0 M
(136.0)180.0 M
(170.0)240.0 M
(204.0)280.0 M
(255.0)280.0 M
(272.0)320.0 M
(306.0)360.0 M
(340.0)420.0 M
(408.0)560.0 M
(510.0)560.0 M(544.0)640.0 M
(612.0)720.0 M
(680.0)900.0 M
(850.0)1400 k1200 k1000 k1800 k(1680)
2400 k(2100)
2400 k(2240)
2800 k1500 k(1440)
1800 k1200 k2000 k(1920)
2400 k1600 k3600 k(3360)
4200 k

3000 k(2880)
3600 k2400 k
3000 k5600 k

4800 k4000 k7200 k(7000)
8400 k6000 k5000 k7200 k6000 k2000 k

600.0 k300.0 k200.0 k720.0 k360.0 k240.0 k1200 k(1050)
1200 k(1120)
1400 k

900.0 k450.0 k300.0 k960.0 k480.0 k320.0 k1200 k

600.0 k400.0 k1800 k(1680)
2400 k(2100)
2800 k1500 k(1440)
1800 k

720.0 k480.0 k900.0 k600.0 k2400 k1200 k

800.0 k3600 k(3500)
4200 k

3000 k1500 k1000 k3600 k1800 k1200 k

×109600 k8000 k5600 k

4800 M

2400 k1600 k

10.00 M

6000 M

3000 k2000 k18.00 M
(16.80)24.00 M
(21.00)24.00 M
(22.40)28.00 M

15.00 M
(14.40)18.00 M

12.00 M

7200 k(7000)
8400 k7200 k3600 k2400 k4500 k

3000 k

16.00 M

12.00 M
(10.50)12.00 M
(11.20)14.00 M

9000 k

20.00 M
(19.20)24.00 M

9600 k

4800 k3200 k12.00 M

6000 k4000 k36.00 M
(33.60)42.00 M

30.00 M
(28.80)36.00 M

24.00 M

7200 k

4800 k30.00 M

15.00 M
(14.40)18.00 M

9000 k6000 k

56.00 M

48.00 M

40.00 M

18.00 M
(16.80)24.00 M
(21.00)28.00 M

24.00 M

12.00 M

8000 k72.00 M
(70.00)84.0 M

60.00 M

50.00 M

30.00 M

15.00 M

10.00 M

72.00 M

60.00 M

36.00 M
(35.00)42.00 M

36.00 M

18.00 M

12.00 M

120.0 M
(112.0)140.0 M

96.00 M

80.00 M

56.00 M

48.00 M

24.00 M

16.00 M

120.0 M

100.0 M

60.00 M

30.00 M

20.00 M

180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

150.0 M
(144.0)180.0 M

120.0 M

72.00 M
(70.00)84.00 M

72.00 M

36.00 M

24.00 M

90.00 M

45.00 M

30.00 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

160.0 M

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M

96.00 M

48.00 M

32.00 M

120.0 M(108.0)120.0 M

56.00 M
(54.00)60.00 M

36.00 M

360.0 M
(336.0)420.0 M

300.0 M
(288.0)360.0 M

240.0 M

150.0 M
(144.0)180.0 M

72.00 M

48.00 M

90.00 M

60.00 M

450.0 M
(448.0)560.0 M
(504.0)560.0 M

400.0 M
(384.0)450.0 M
(432.0)480.0 M

320.0 M

180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

96.00 M

64.00 M

120.0 M(108.0)120.0 M

72.00 M

720.0 M
(700.0)840.0 M

600.0 M

500.0 M

300.0 M

150.0 M

100.0 M

720.0 M

600.0 M

360.0 M
(350.0)420.0 M

360.0 M

180.0 M

120.0 M

960.0 M

800.0 M

560.0 M

480.0 M

240.0 M

160.0 M

1000 M

720.0 M
(700.0)840.0 M

600.0 M

300.0 M

200.0 M

720.0 M

360.0 M

240.0 M

900.0 M

450.0 M

300.0 M

960.0 M

480.0 M

320.0 M

560.0 M
(540.0)600.0 M

360.0 M

12kA

720.0 M

480.0 M

15kA

900.0 M

75A

50.00 M
60.00 M
75.00 M

80A

80.00 M

100A

100.0 M

120A

120.0 M

150A

150.0 M

200A
250A
300A

200.0 M
250.0 M
300.0 M

400A

400.0 M

500A

500.0 M

600A

600.0 M

750A
800A
900A
1000A
1200A

1800A
2000A
2500A
3000A

750.0 M
800.0 M
900.0 M
1000 M

32.00 M

120.0 M

320.0 M

1000 M

12.00 M
(11.20)14.00 M

1500 k

720.0 k(700.0)840 k

12.00 M

60A

40.00 M

2500 k
3000 k25.00 M

50A

×100008000 k

12A

40A

×1000

7500 k3600 k(3455)
4200 k(4145)
5600 M(5182)
5600 k(5527)
7200 k(6909)
8400 k(8291)
12.00 M
(10.36)14.00 M
(13.82)18.00 M
(17.27)24.00 M
(20.73)28.00 M
(27.64)36.00 M
(34.55)42.00 M
(41.45)56.00 M
(51.82)56.00 M
(55.27)72.00 M
(69.09)84.00 M
(82.91)120.0 M
(103.6)140.0 M(138.2)180.0 M
(172.7)240.0 M
(207.3)280.0 M
(276.4)360.0 M
(345.5)420.0 M
(414.5)560.0 M
(518.2)560.0 M
(552.7)640.0 M
(621.8)720.0 M
(690.9)840.0 M
(829.1)

4000A
5000A
6000A
7500A
8000A
9000A
10kA

15.00 M

20.00 M

150.0 M

180.0 M
200.0 M

300.0 M

360.0 M
400.0 M×100

40.00 M

80.00 M×1000

400.0 M

600.0 M
800.0 M

20kA

×1000030kA<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW62<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173
Appendix table 1-2
Active power range, watt-hour multiplier rate table (Three-phase three-wire)
25kV
9000V
4500V
24kV
18kV
18kV
15kV
3000V
2400V
1500V
1200V
600V
V range(VT18.4kV/110V) (VT16.5kV/110V) (VT13.8kV/110V) (VT13.2kV/110V) (VT11kV/110V) (VT6600/110V) (VT3300/110V) (VT2200/110V) (VT1650/110V) (VT1100/110V) (VT880/110V) (VT480/110V) Multiplying
Multiplying
factorfactor
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
A range
5A
6A
7.5A
8A
10A
12A
15A
20A
×10

25A
30A
40A
50A
60A
75A
80A
100A
120A
150A
200A

×100

250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A

×1000

2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10kA
12kA
15kA

×1000020kA
30kA

180.0 k(167.3)240.0 k(200.7)280.0 k(250.9)280.0 k(267.6)360.0 k(334.5)420.0 k(401.5)560.0 k(501.8)720.0 k(669.1)840.0 k(836.4)1200 k(1004)
1400 k(1338)
1800 k(1673)
2400 k(2007)
2800 k(2509)
2800 k(2676)
3600 k(3345)
4200 k(4015)
5600 k(5018)
7200 k(6691)
8400 k(8364)
12.00 M
(10.04)14.00 M
(13.38)18.00 M
(16.73)24.00 M
(20.07)28.00 M
(25.09)28.00 M
(26.76)32.00 M
(30.11)36.00 M
(33.45)42.00 M
(40.15)56.00 M
(50.18)56.00 M
(53.53)64.00 M
(60.22)72.00 M
(66.91)84.00 M
(83.64)120.0 M
(100.4)140.0 M
(133.8)180.0 M(167.3)240.0 M
(200.7)280.0 M
(250.9)280.0 M
(267.6)320.0 M
(301.1)360.0 M
(334.5)420.0 M
(401.5)560.0 M
(501.8)720.0 M
(669.1)150.0 k180.0 k240.0 k(225.0)240.0 k300.0 k360.0 k450.0 k600.0 k750.0 k900.0 k1200 k1500 k1800 k2400 k(2250)
2400 k
3000 k3600 k4500 k6000 k7500 k9000 k12.00 M
15.00 M
18.00 M
24.00 M
(22.50)24.00 M
28.00 M
(27.00)30.00 M
36.00 M
45.00 M
48.00 M
56.00 M
(54.00)60.00 M
75.00 M
90.00 M
120.0 M
150.0 M
180.0 M
240.0 M
(225.0)240.0 M
280.0 M
(270.0)300.0 M
360.0 M
450.0 M
600.0 M
900.0 M

140.0 k(125.5)160.0 k(150.5)200.0 k(188.2)240.0 k(200.7)280.0 k(250.9)320.0 k(301.1)400.0 k(376.4)560.0 k(501.8)640.0 k(627.3)800.0 k(752.7)1200 k(1004)
1400 k(1255)
1600 k(1505)
2000 k(1882)
2400 k(2007)
2800 k(2509)
3200 k(3011)
4000 k(3764)
5600 k(5018)
6400 k(6273)
8000 k(7527)
12.00 M
(10.04)14.00 M
(12.55)16.00 M
(15.05)20.00 M(18.82)24.00 M
(20.07)24.00 M
(22.58)28.00 M
(25.09)32.00 M
(30.11)40.00 M(37.64)42.00 M
(40.15)48.00 M
(45.16)56.00 M
(50.18)64.00 M
(62.73)80.00 M
(75.27)120.0 M
(100.4)140.0 M
(125.5)160.0 M
(150.5)200.0 M
(188.2)240.0 M
(200.7)240.0 M
(225.8)280.0 M
(250.9)320.0 M
(301.1)400.0 M
(376.4)560.0 M
(501.8)800.0 M
(752.7)120.0 k100.0 k60.00 k30.00 k20.00 k15.00 k10.00 k8000

150.0 k(144.0)180.0 k120.0 k72.00 k36.00 k24.00 k18.00 k12.00 k9600

150.0 k90.00 k45.00 k30.00 k12.00 k160.0 k96.00 k48.00 k32.00 k24.00 k(22.50)24.00 k15.00 k200.0 k(192.0)240.0 k16.00 k200.0 k120.0 k60.00 k40.00 k30.00 k20.00 k14.00 k(12.80)16.00 k300.0 k(288.0)360.0 k240.0 k72.00 k48.00 k36.00 k24.00 k300.0 k150.0 k(144.0)180.0 k90.00 k60.00 k45.00 k30.00 k20.00 k(19.20)24.00 k480.0 k400.0 k240.0 k120.0 k80.00 k60.00 k40.00 k32.00 k600.0 k500.0 k300.0 k150.0 k100.0 k75.00 k50.00 k40.00 k720.0 k600.0 k360.0 k180.0 k120.0 k90.00 k60.00 k48.00 k960.0 k800.0 k480.0 k240.0 k160.0 k120.0 k80.00 k64.00 k1200 k1000 k

600.0 k300.0 k200.0 k150.0 k100.0 k80.00 k1500 k(1440)
1800 k1200 k

720.0 k360.0 k240.0 k180.0 k120.0 k96.00 k1500 k

900.0 k450.0 k300.0 k120.0 k1600 k

960.0 k480.0 k320.0 k240.0 k(225.0)240.0 k150.0 k2000 k(1920)
2400 k

160.0 k2000 k1200 k

600.0 k400.0 k300.0 k200.0 k140.0 k(128.0)160.0 k3000 k(2880)
3600 k2400 k

720.0 k480.0 k360.0 k240.0 k3000 k1500 k(1440)
1800 k

900.0 k600.0 k450.0 k300.0 k200.0 k(192.0)240.0 k

4800 k4000 k2400 k1200 k

800.0 k600.0 k400.0 k320.0 k6000 k5000 k

3000 k1500 k1000 k

750.0 k500.0 k400.0 k7200 k6000 k3600 k1800 k1200 k

900.0 k600.0 k480.0 k9600 k8000 k

4800 k2400 k1600 k1200 k

800.0 k640.0 k12.00 M

10.00 M

6000 k

3000 k2000 k1500 k1000 k

800.0 k15.00 M
(14.40)18.00 M

12.00 M

7200 k3600 k2400 k1800 k1200 k

960.0 k15.00 M

9000 k4500 k

3000 k1500 k1200 k

20.00 M
(19.20)24.00 M
(21.60)24.00 M

16.00 M

9600 k

4800 k3200 k2400 k(2250)
2400 k1600 k18.00 M

12.00 M
(10.80)12.00 M

5600 k(5400)
6000 k3600 k1400 k(1280)
1500 k(1440)
1600 k30.00 M
(28.80)36.00 M

24.00 M

15.00 M
(14.40)18.00 M

7200 k

40.00 M
(38.40)45.00 M(43.20)48.00 M

32.00 M

60.00 M

20.00 M

30.00 M

1800 k4000 k2800 k(2700)
3000 k

4800 k3600 k2400 k9000 k6000 k4500 k

3000 k2000 k2000 k(1920)
2400 k9600 k6400 k

4800 k3200 k12.00 M
(10.80)12.00 M

7200 k8000 k5600 k(5400)
6000 k3600 k

40.00 M

20.00 M
(19.20)24.00 M
(21.60)24.00 M

4000 k2800 k(2560)
3000 k(2880)
3200 k

50.00 M

30.00 M

15.00 M

10.00 M

7500 k5000 k4000 k72.00 M

60.00 M

36.00 M

18.00 M

12.00 M

9000 k6000 k

4800 k

96.00 M

80.00 M

48.00 M

24.00 M

16.00 M

12.00 M

8000 k6400 k

120.0 M

100.0 M

60.00 M

30.00 M

20.00 M

15.00 M

10.00 M

8000 k

150.0 M
(144.0)180.0 M

120.0 M

72.00 M

36.00 M

24.00 M

18.00 M

12.00 M

9600 k

150.0 M

90.00 M

45.00 M

30.00 M

12.00 M

160.0 M

96.00 M

48.00 M

32.00 M

24.00 M
(22.50)24.00 M

15.00 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

16.00 M

180.0 M

120.0 M(108.0)120.0 M

56.00 M
(54.00)60.00 M

36.00 M

28.00 M
(27.00)30.00 M

18.00 M

40.00 M

14.00 M(12.80)15.00 M
(14.40)16.00 M

300.0 M
(288.0)360.0 M

240.0 M

150.0 M
(144.0)180.0 M

72.00 M

48.00 M

36.00 M

24.00 M

90.00 M

60.00 M

45.00 M

30.00 M

36.00 M

200.0 M

300.0 M

20.00 M

20.00 M
(19.20)24.00 M

480.0 M

400.0 M

240.0 M

120.0 M

80.00 M

60.00 M

40.00 M

32.00 M

720.0 M

600.0 M

360.0 M

180.0 M

120.0 M

90.00 M

60.00 M

48.00 M

<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW

63

4500
(4364)
5600(5236)
7200(6545)
7200(6982)
9000(8727)
12.00 k(10.47)14.00 k(13.09)18.00 k(17.45)24.00 k(21.82)28.00 k(26.18)36.00 k(34.91)45.00 k(43.64)56.00 k(52.36)72.00 k(65.45)72.00 k(69.82)90.00 k(87.27)120.0 k(104.7)140.0 k(130.9)180.0 k(174.5)240.0 k(218.2)280.0 k(261.8)360.0 k(349.1)450.0 k(436.4)560.0 k(523.6)720.0 k(654.5)720.0 k(698.2)800.0 k(785.5)900.0 k(872.7)1200 k(1047)
1400 k(1309)
1400 k(1396)
1600 k(1571)
1800 k(1745)
2400 k(2182)
2800 k(2618)
3600 k(3491)
4500 k(4364)
5600 k(5236)
7200 k(6545)
7200 k(6982)
8000 k(7855)
9000 k(8727)
12.00 M
(10.47)14.00 M
(13.09)18.00 M
(17.45)28.00 M
(26.18)

×0.1

×1

×10

×100

×1000

<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173
Appendix table 1-3
Active power range, watt-hour multiplier rate table (Three-phase three-wire)
600V
600V
500V
300V
V range(VT460/110V) (VT440/110V) (VT380/110V) (VT220/110V)
Multiplying
factor
[W]
[W]
[W]
[W]
A range
5A
6A
7.5A
8A
10A
×0.1

12A
15A
20A
25A
30A
40A
50A
60A
75A
80A
100A

×1

120A
150A
200A
250A
300A
400A
500A
600A
750A
800A
900A
1000A

×10

1200A
1500A
1600A
1800A
2000A
2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10kA

×100

12kA
15kA
20kA

×1000

30kA

4200
(4182)
5600(5018)
6400(6273)
7200(6691)
8400(8364)
12.00 k(10.04)14.00 k(12.55)18.00 k(16.73)24.00 k(20.91)28.00 k(25.09)36.00 k(33.45)42.00 k(41.82)56.00 k(50.18)64.00 k(62.73)72.00 k(66.91)84.00 k(83.64)120.0 k(100.4)140.0 k(125.5)180.0 k(167.3)240.0 k(209.1)280.0 k(250.9)360.0 k(334.5)420.0 k(418.2)560.0 k(501.8)640.0 k(627.3)720.0 k(669.1)800.0 k(752.7)840.0 k(836.4)1200 k(1004)
1400 k(1255)
1400 k(1338)
1600 k(1505)
1800 k(1673)
2400 k(2091)
2800 k(2509)
3600 k(3345)
4200 k(4182)
5600 k(5018)
6400 k(6273)
7200 k(6691)
8000 k(7527)
8400 k(8364)
12.00 M
(10.04)14.00 M
(12.55)18.00 M
(16.73)28.00 M
(25.09)

4000
4800
6000
6400
8000
9600
12.00 k16.00 k20.00 k24.00 k32.00 k40.00 k48.00 k60.00 k64.00 k80.00 k96.00 k120.0 k160.0 k200.0 k240.0 k320.0 k400.0 k480.0 k600.0 k640.0 k720.0 k800.0 k960.0 k1200 k1400 k(1280)
1500 k(1440)
1600 k2000 k2400 k3200 k4000 k
4800 k6000 k6400 k7200 k8000 k9600 k12.00 M
16.00 M
24.00 M

3600
(3455)
4200(4145)
5600(5182)
5600(5527)
7200(6909)
8400(8291)
12.00 k(10.36)14.00 k(13.82)18.00 k(17.27)24.00 k(20.73)28.00 k(27.64)36.00 k(34.55)42.00 k(41.45)56.00 k(51.82)56.00 k(55.27)72.00 k(69.09)84.00 k(82.91)120.0 k(103.6)140.0 k(138.2)180.0 k(172.7)240.0 k(207.3)280.0 k(276.4)360.0 k(345.5)420.0 k(414.5)560.0 k(518.2)560.0 k(552.7)640.0 k(621.8)720.0 k(690.9)840.0 k(829.1)1200 k(1036)
1200 k(1105)
1400 k(1244)
1400 k(1382)
1800 k(1727)
2400 k(2073)
2800 k(2764)
3600 k(3455)
4200 k(4145)
5600 k(5182)
5600 k(5527)
6400 k(6218)
7200 k(6909)
8400 k(8291)
12.00 M
(10.36)14.00 M
(13.82)24.00 M
(20.73)150V(110V)
[W]

2000

1000

2400

1200

3000

1500

3200

1600

4000

2000

4800

2400

6000

3000

Multiplying
factor
×0.01

8000

4000

10.00 k5000

12.00 k6000

16.00 k8000

20.00 k10.00 k24.00 k12.00 k30.00 k15.00 k32.00 k16.00 k40.00 k20.00 k48.00 k24.00 k60.00 k30.00 k80.00 k40.00 k100.0 k50.00 k120.0 k60.00 k160.0 k80.00 k200.0 k100.0 k240.0 k120.0 k300.0 k150.0 k320.0 k160.0 k360.0 k180.0 k400.0 k200.0 k480.0 k240.0 k600.0 k300.0 k640.0 k320.0 k720.0 k360.0 k800.0 k400.0 k1000 k

500.0 k1200 k

600.0 k1600 k

800.0 k2000 k1000 k2400 k1200 k

3000 k1500 k3200 k1600 k3600 k1800 k4000 k2000 k

4800 k2400 k6000 k

3000 k8000 k4000 k12.00 M

6000 k

×0.1

×1

×10

×100

<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW

64<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173
Appendix table 2
Active power range, watt-hour multiplier rate table (Single-phase 3-wire)
V range

150V(110V)[W]

A range
5A
6A
7.5A
8A
10A
12A
15A
20A
25A
30A
40A
50A
60A
75A
80A
100A
120A
150A
200A
250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A
2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A
20000A
30000A

Multiplying
factor

1000
1200
1500
1600
2000
2400
3000
4000
5000
6000
8000
×0.1
10.00 k12.00 k15.00 k16.00 k20.00 k24.00 k30.00 k40.00 k50.00 k60.00 k80.00 k×1
100.0 k120.0 k150.0 k160.0 k180.0 k200.0 k240.0 k300.0 k320.0 k360.0 k400.0 k500.0 k600.0 k800.0 k×101000 k1200 k1500 k1600 k1800 k2000 k2400 k
3000 k4000 k6000 k×100<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW

65<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173
Appendix table 3-1
Active power range, watt-hour multiplier rate table (Single-phase two-wire)
105kV
90kV
300kV
255kV
210kV
750kV
500kV
180kV
150kV
45kV
30kV
375kV
V range(VT550kV/110V) (VT380kV/110V) (VT275kV/110V) (VT220kV/110V) (VT187kV/110V) (VT154kV/110V) (VT132kV/110V) (VT110kV/110V) (VT77kV/110V) (VT66kV/110V) (VT33kV/110V) (VT22kV/110V) Multiplying
Multiplying
factorfactor
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
A range
[W]
[W]
[W]

×100

×1000

5A

2500 k

6A

3000 k

7.5A
8A

4000 k(3750)
4000 k

10A

5000 k

12A

6000 k15A

7500 k

20A

10.00 M

25A
30A

14.00 M
(12.50) M
15.00 M

40A

20.00 M

50A

25.00 M

60A

30.00 M

75A
80A

40.00 M(37.50)40.00 M

100A

50.00 M

120A

60.00 M

150A

75.00 M

200A

100.0 M

250A
300A

140.0 M
(125.0) M
150.0 M

400A

200.0 M

500A

250.0 M

600A

300.0 M

750A

400.0 M
(375.0) M
400.0 M

800A
900A
1000A

500.0 M

1200A

600.0 M

1500A

750.0 M

1600A
×10000450.0 M

1800A
2000A
2500A
3000A
4000A
5000A
6000A

800.0 M
900.0 M

1800 k(1727)
2400 k(2073)
2800 k(2591)
2800 k(2764)
3600 k(3455)
4200 k(4145)
5600 k(5182)
7200 k(6909)
9000 k(8636)
12.00 M
(10.36)14.00 M
(13.82)18.00 M
(17.27)24.00 M
(20.73)28.00 M
(25.91)28.00 M
(27.64)36.00 M
(34.55)42.00 M
(41.45)56.00 M
(51.82)72.00 M
(69.09)90.00 M
(86.36)120.0 M
(103.6)140.0 M(138.2)180.0 M
(172.7)240.0 M
(207.3)280.0 M
(259.1)280.0 M
(276.4)320.0 M
(310.9)360.0 M
(345.5)420.0 M
(414.5)560.0 M
(518.2)560.0 M
(552.7)640.0 M
(621.8)720.0 M
(690.9)900.0 M
(863.6)

1400 k(1250)
1500 k1000 k2000 k(1875)
2000 k1500 k2500 k2000 k

3000 k2400 k4000 k(3750)
5000 k

3000 k6400 k(6250)
7500 M

5000 k

10.00 M

8000 k

14.00 M
(12.50)15.00 M

10.00 M

1200 k1600 k4000 k6000 k12.00 M

20.00 M
(18.75)20.00 M

15.00 M

25.00 M

20.00 M

30.00 M

24.00 M

40.00 M(37.50)50.00 M

40.00 M

16.00 M

30.00 M

64.00 M
(62.50)75.00 M

50.00 M

100.0 M

80.00 M

140.0 M
(125.0)150.0 M

100.0 M

60.00 M

120.0 M

200.0 M
(187.5)200.0 M

150.0 M

240.0 M
(225.0)250.0 M

180.0 M
200.0 M

300.0 M

240.0 M

400.0 M
(375.0)400.0 M

300.0 M
320.0 M

450.0 M

360.0 M

500.0 M

400.0 M

640.0 M
(625.0)750.0 M

500.0 M

160.0 M

600.0 M
800.0 M

900.0 k(850.0)1200 k(1020)
1400 k(1275)
1400 k(1360)
1800 k(1700)
2400 k(2040)
2800 k(2550)
3600 k(3400)
4500 k(4250)
5600 k(5100)
7200 k(6800)
9000 k(8500)
12.00 M
(10.20)14.00 M(12.75)14.00 M
(13.60)18.00 M
(17.00)24.00 M
(20.40)28.00 M
(25.50)36.00 M
(34.00)45.00 M
(42.50)56.00 M
(51.00)72.00 M
(68.00)90.00 M
(85.00)120.0 M
(102.0)140.0 M
(127.5)140.0 M
(136.0)160.0 M
(153.0)180.0 M
(170.0)240.0 M
(204.0)280.0 M
(255.0)280.0 M
(272.0)320.0 M
(306.0)360.0 M
(340.0)450.0 M
(425.0)560.0 M
(510.0)720.0 M
(680.0)900.0 M
(850.0)720.0 k(700.0)840.0 k600.0 k500.0 k720.0 k600.0 k1200 k(1050)
1200 k(1120)
1400 k

900.0 k750.0 k960.0 k800.0 k1200 k1000 k1800 k(1680)
2400 k(2100)
2800 k1500 k(1440)
1800 k1200 k2400 k2000 k3600 k(3500)
4200 k

3000 k2500 k3600 k

3000 k5600 M

4800 k4000 k7200 M(7000)
8400 M

6000 k5000 k7200 k6000 k9000 k7500 k9600 k8000 k12.00 M

10.00 M

18.00 M
(16.80)24.00 M
(21.00)28.00 M

15.00 M
(14.40)18.00 M

12.00 M
15.00 M

24.00 M

20.00 M

36.00 M
(35.00)42.00 M

30.00 M

25.00 M

36.00 M

30.00 M

56.00 M

48.00 M

40.00 M

72.00 M
(70.00)84.00 M

60.00 M

50.00 M

1500 k

360.0 k(350.0)420.0 k300.0 k150.0 k100.0 k360.0 k180.0 k120.0 k560.0 k(525.0)560.0 k450.0 k150.0 k480.0 k240.0 k(225.0) k
240.0 k720.0 k(700.0)840.0 k600.0 k300.0 k200.0 k720.0 k360.0 k240.0 k1200 k(1050)
1400 k

900.0 k450.0 k300.0 k1200 k

600.0 k400.0 k1800 k(1750)
2400 k(2100)
2800 k1500 k

750.0 k500.0 k1800 k

900.0 k600.0 k2400 k1200 k

800.0 k3000 k1500 k1000 k3600 k1800 k1200 k5600 k(5250)
5600 k4500 k2400 k(2250)
2400 k1500 k

4800 k7200 k(7000)
8400 k6000 k

3000 k2000 k7200 k3600 k2400 k12.00 M
(10.50)14.00 M

9000 M

4500 k

3000 k12.00 M

6000 k4000 k

15.00 M

7500 k5000 k18.00 M

9000 k6000 k

24.00 M

12.00 M

8000 k30.00 M

15.00 M

10.00 M

160.0 k×10

12.00 M
(10.50)12.00 M
(11.20)14.00 M

3600 k(3500)
4200 k18.00 M
(17.50)24.00 M
(21.00)28.00 M

1600 k×100

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M
180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

72.00 M

60.00 M

90.00 M

75.00 M

96.00 M

80.00 M

120.0 M(108.0)120.0 M

90.00 M

150.0 M
(144.0)180.0 M

100.0 M
120.0 M
150.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

160.0 M

200.0 M

360.0 M
(350.0)420.0 M

300.0 M

250.0 M

360.0 M

300.0 M

560.0 M

480.0 M

400.0 M

720.0 M
(700.0)840.0 M

600.0 M

500.0 M

720.0 M

600.0 M

900.0 M

750.0 M

180.0 M

36.00 M
(35.00)42.00 M

36.00 M

18.00 M

12.00 M

56.00 M(52.50)56.00 M

45.00 M

15.00 M

48.00 M

24.00 M
(22.50)24.00 M

64.00 M
(63.00)72.00 M
(70.00)84.00 M

56.00 M
(54.00) M
60.00 M

28.00 M
(27.00)30.00 M

18.00 M
20.00 M

72.00 M

36.00 M

24.00 M

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M

90.00 M

45.00 M

30.00 M

16.00 M

96.00 M

48.00 M

32.00 M

120.0 M(108.0) M
120.0 M

56.00 M
(54.00)60.00 M

36.00 M

180.0 M
(175.0)240.0 M
(210.0)280.0 M

150.0 M

75.00 M

50.00 M

180.0 M

90.00 M

60.00 M

240.0 M

120.0 M

80.00 M

360.0 M
(350.0)420.0 M

300.0 M

150.0 M

100.0 M

360.0 M

180.0 M

120.0 M

560.0 M
(525.0)560.0 M

450.0 M

240.0 M
(225.0) M
240.0 M

150.0 M

480.0 M

640.0 M
(630.0)720.0 M
(700.0)840.0 M

560.0 M
(540.0) M
600.0 M

280.0 M
(270.0) M
300.0 M

40.00 M

×1000

7500A

800.0 M

8000A

900.0 M

9000A
10kA

160.0 M
180.0 M
200.0 M

720.0 M

360.0 M

240.0 M

900.0 M

450.0 M

300.0 M

20kA

600.0 M

400.0 M

30kA

900.0 M

600.0 M

12kA
15kA

×10000<Note 1>
Parenthesis is primary active power (reactive power) value in /500W(500var).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW

66<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173
Appendix table 3-2
Active power range, watt-hour multiplier rate table (Single-phase two-wire)
24kV
18kV
18kV
15kV
3000V
2400V
1500V
1200V
600V
9000V
4500V
25kV
V range(VT18.4kV/110V) (VT16.5kV/110V) (VT13.8kV/110V) (VT13.2kV/110V) (VT11kV/110V) (VT6600/110V) (VT3300/110V) (VT2200/110V) (VT1650/110V) (VT1100/110V) (VT880/110V) (VT480/110V) Multiplying
Multiplying
factorfactor
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
A range
[W]
×1

5A
6A
7.5A
8A
10A
12A
15A
20A
25A
30A

×10

40A
50A
60A
75A
80A
100A
120A
150A
200A
250A
300A

×100

400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A
2500A
3000A

×1000

4000A
5000A
6000A
7500A
8000A
9000A
10kA
12kA
15kA
20kA

×1000030kA

84.00 k(83.64)120.0 k(100.4)140.0 k(125.5)140.0 k(133.8)180.0 k(167.3)240.0 k(200.7)280.0 k(250.9)360.0 k(334.5)420.0 k(418.2)560.0 k(501.8)720.0 k(669.1)840.0 k(836.4)1200 k(1004)
1400 k(1255)
1400 k(1338)
1800 k(1673)
2400 k(2007)
2800 k(2509)
3600 k(3345)
4200 k(4182)
5600 k(5018)
7200 k(6691)
8400 k(8364)
12.00 M
(10.04)14.00 M
(12.55)14.00 M
(13.38)16.00 M
(15.05)18.00 M
(16.73)24.00 M
(20.07)28.00 M
(25.09)28.00 M
(26.76)32.00 M
(30.11)36.00 M
(33.45)42.00 M
(41.82)56.00 M
(50.18)72.00 M
(66.91)84.00 M
(83.64)120.0 M
(100.4)140.0 M
(125.5)140.0 M
(133.8)160.0 M
(150.5)180.0 M(167.3)240.0 M
(200.7)280.0 M
(250.9)360.0 M
(334.5)560.0 M
(501.8)75.00 k90.00 k120.0 k(112.5)120.0 k150.0 k180.0 k240.0 k(225.0)300.0 k400.0 k(375.0)450.0 k600.0 k750.0 k900.0 k1200 k(1125)
1200 k1500 k1800 k2400 k(2250)
3000 k4000 k(3750)
4500 k6000 k7500 k9000 k12.00 M
(11.25)12.00 M
14.00 M
(13.50)15.00 M
18.00 M
24.00 M
(22.50)24.00 M
28.00 M
(27.00)30.00 M
40.00 M(37.50)45.00 M
60.00 M
75.00 M
90.00 M
120.0 M
(112.5)120.0 M
140.0 M
(135.0)150.0 M
180.0 M
240.0 M
(225.0)300.0 M
450.0 M

64.00 k(62.73)80.00 k(75.27)96.00 k(94.09)120.0 k(100.4)140.0 k(125.5)160.0 k(150.5)200.0 k(188.2)280.0 k(250.9)320.0 k(313.6)400.0 k(376.4)560.0 k(501.8)640.0 k(627.3)800.0 k(752.7)960.0 k(940.9)1200 k(1004)
1400 k(1255)
1600 k(1505)
2000 k(1882)
2800 k(2509)
3200 k(3136)
4000 k(3764)
5600 k(5018)
6400 k(6273)
8000 k(7527)
9600 k(9409)
12.00 M
(10.04)12.00 M
(11.29)14.00 M
(12.55)16.00 M
(15.05)20.00 M(18.82)24.00 M
(20.07)24.00 M
(22.58)28.00 M
(25.09)32.00 M
(31.36)40.00 M(37.64)56.00 M
(50.18)64.00 M
(62.73)80.00 M
(75.27)96.00 M
(94.09)120.0 M
(100.4)120.0 M
(112.9)140.0 M
(125.5)160.0 M
(150.5)200.0 M
(188.2)280.0 M
(250.9)400.0 M
(376.4)60.00 k50.00 k30.00 k15.00 k10.00 k

7500

5000

4000

72.00 k60.00 k36.00 k18.00 k12.00 k

9000

6000

4800

90.00 k75.00 k45.00 k6000

48.00 k16.00 k12.00 k(11.25)12.00 k7500

80.00 k24.00 k(22.50) k
24.00 k15.00 k96.00 k

8000

6400

120.0 k100.0 k60.00 k30.00 k20.00 k15.00 k10.00 k8000

150.0 k(144.0)180.0 k120.0 k72.00 k36.00 k24.00 k18.00 k12.00 k9600

150.0 k90.00 k45.00 k30.00 k15.00 k12.00 k240.0 k200.0 k120.0 k60.00 k40.00 k24.00 k(22.50)30.00 k20.00 k16.00 k300.0 k250.0 k150.0 k75.00 k50.00 k360.0 k300.0 k180.0 k90.00 k480.0 k400.0 k240.0 k120.0 k600.0 k500.0 k300.0 k720.0 k600.0 k900.0 k960.0 k25.00 k20.00 k60.00 k40.00 k(37.50)45.00 k30.00 k24.00 k80.00 k60.00 k40.00 k32.00 k150.0 k100.0 k75.00 k50.00 k40.00 k360.0 k180.0 k120.0 k90.00 k60.00 k48.00 k750.0 k450.0 k60.00 k160.0 k120.0 k(112.5)120.0 k75.00 k480.0 k240.0 k(225.0)240.0 k150.0 k800.0 k80.00 k64.00 k1200 k1000 k

600.0 k300.0 k200.0 k150.0 k100.0 k80.00 k1500 k(1440)
1800 k1200 k

720.0 k360.0 k240.0 k180.0 k120.0 k96.00 k1500 k

900.0 k450.0 k300.0 k120.0 k2000 k1200 k

600.0 k400.0 k240.0 k(225.0)300.0 k150.0 k2400 k

200.0 k160.0 k3000 k2500 k1500 k

750.0 k500.0 k250.0 k200.0 k3600 k

3000 k1800 k

900.0 k600.0 k400.0 k(375.0)450.0 k300.0 k240.0 k

4800 k4000 k2400 k1200 k

800.0 k600.0 k400.0 k320.0 k6000 k5000 k

3000 k1500 k1000 k

750.0 k500.0 k400.0 k7200 k6000 k3600 k1800 k1200 k

900.0 k600.0 k480.0 k9000 k7500 k4500 k1500 k1200 k(1125)
1200 k

750.0 k600.0 k800.0 k640.0 k900.0 k720.0 k9600 k8000 k

4800 k2400 k(2250)
2400 k12.00 M
(10.80)12.00 M

9000 k2800 k(2700) k
3000 k1800 k1600 k

10.00 M

5600 k(5400)
6000 k2000 k1400 k(1350)
1500 k1000 k

800.0 k15.00 M
(14.40)18.00 M

12.00 M

7200 k3600 k2400 k1800 k1200 k

960.0 k15.00 M

9000 k4500 k

3000 k1500 k1200 k

20.00 M
(19.20)24.00 M
(21.60)24.00 M

16.00 M

9600 k

4800 k3200 k2400 k(2250)
2400 k1600 k18.00 M

5600 k(5400)
6000 k3600 k2800 k(2700)
3000 k1800 k

20.00 M

12.00 M
(10.80)12.00 M

2000 k1400 k(1280)
1500 k(1440)
1600 k30.00 M

25.00 M

15.00 M

7500 k5000 k2500 k2000 k4000 k36.00 M

30.00 M

18.00 M

9000 k6000 k4000 k(3750)
4500 k

3000 k2400 k

48.00 M

40.00 M

24.00 M

12.00 M

8000 k6000 k4000 k3200 k

60.00 M

50.00 M

30.00 M

15.00 M

10.00 M

7500 k5000 k4000 k72.00 M

60.00 M

36.00 M

18.00 M

12.00 M

9000 k6000 k

4800 k

90.00 M

75.00 M

45.00 M

6000 k

48.00 M

12.00 M
(11.25)12.00 M

7500 k80.00 M

24.00 M
(22.50)24.00 M

15.00 M

96.00 M

8000 k6400 k

120.0 M(108.0)120.0 M

90.00 M

28.00 M
(27.00)30.00 M

18.00 M

7200 k

20.00 M

14.00 M
(13.50)15.00 M

9000 k

100.0 M

56.00 M
(54.00)60.00 M

10.00 M

8000 k

150.0 M
(144.0)180.0 M

120.0 M

72.00 M

36.00 M

24.00 M

18.00 M

12.00 M

9600 k

150.0 M

90.00 M

45.00 M

30.00 M

15.00 M

12.00 M

240.0 M

200.0 M

120.0 M

60.00 M

40.00 M

24.00 M
(22.50)30.00 M

20.00 M

16.00 M

360.0 M

300.0 M

180.0 M

90.00 M

60.00 M

45.00 M

30.00 M

24.00 M

<Note 1>
Parenthesis is primary active power (reactive power) value in /500W(500var).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW

67

16.00 M

2400
(2182)
2800(2618)
3600(3273)
3600(3491)
4500(4364)
5600(5236)
7200(6545)
9000(8727)
12.00 k(10.91)14.00 k(13.09)18.00 k(17.45)24.00 k(21.82)28.00 k(26.18)36.00 k(32.73)36.00 k(34.91)45.00 k(43.64)56.00 k52.36
72.00 k(65.45)90.00 k(87.27)120.0 k(109.1)140.0 k(130.9)180.0 k(174.5)240.0 k(218.2)280.0 k(261.8)360.0 k(327.3)360.0 k(349.1)400.0 k(392.7)450.0 k(436.4)560.0 k(523.6)720.0 k(654.5)720.0 k(698.2)800.0 k(785.5)900.0 k(872.7)1200 k(1091)
1400 k(1309)
1800 k(1745)
2400 k(2182)
2800 k(2618)
3600 k(3273)
3600 k(3491)
4000 k(3927)
4500 k(4364)
5600 k(5236)
7200 k(6545)
9000 k(8727)
14.00 M
(13.09)

×0.1

×1

×10

×100
×1000

<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173
Appendix table 3-3
Active power range, watt-hour multiplier rate table (Single-phase two-wire)
600V
600V
500V
300V
V range(VT460/110V) (VT440/110V) (VT380/110V) (VT220/110V)
Multiplying
factor[W][W]
A range
[W]
[W]
5A
6A
7.5A
8A
10A
12A
15A
×0.1

20A
25A
30A
40A
50A
60A
75A
80A
100A
120A
150A

×1

200A
250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A

×10

2000A
2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10kA
12kA
15kA

×100
×1000

20kA
30kA

2400
(2091)
2800(2509)
3200(3136)
3600(3345)
4200(4182)
5600(5018)
6400(6273)
8400(8364)
12.00 k(10.45)14.00 k(12.55)18.00 k(16.73)24.00 k(20.91)28.00 k(25.09)32.00 k(31.36)36.00 k(33.45)42.00 k(41.82)56.00 k(50.18)64.00 k(62.73)84.00 k(83.64)120.0 k(104.5)140.0 k(125.5)180.0 k(167.3)240.0 k(209.1)280.0 k(250.9)320.0 k(313.6)360.0 k(334.5)400.0 k(376.4)420.0 k(418.2)560.0 k(501.8)640.0 k(627.3)720.0 k(669.1)800.0 k(752.7)840.0 k(836.4)1200 k(1045)
1400 k(1255)
1800 k(1673)
2400 k(2091)
2800 k(2509)
3200 k(3136)
3600 k(3345)
4000 k(3764)
4200 k(4182)
5600 k(5018)
6400 k(6273)
8400 k(8364)
14.00 M
(12.55)

2000
2400
3000
3200
4000
4800
6000
8000
10.00 k12.00 k16.00 k20.00 k24.00 k30.00 k32.00 k40.00 k48.00 k60.00 k80.00 k100.0 k120.0 k160.0 k200.0 k240.0 k300.0 k320.0 k360.0 k400.0 k480.0 k600.0 k640.0 k720.0 k800.0 k1000 k1200 k1600 k2000 k2400 k
3000 k3200 k3600 k4000 k
4800 k6000 k8000 k12.00 M

1800
(1727)
2400(2073)
2800(2591)
2800(2764)
3600(3455)
4200(4145)
5600(5182)
7200(6909)
9000(8636)
12.00 k(10.36)14.00 k(13.82)18.00 k(17.27)24.00 k(20.73)28.00 k(25.91)28.00 k(27.64)36.00 k(34.55)42.00 k(41.45)56.00 k(51.82)72.00 k(69.09)90.00 k(86.36)120.0 k(103.6)140.0 k(138.2)180.0 k(172.7)240.0 k(207.3)280.0 k(259.1)280.0 k(276.4)320.0 k(310.9)360.0 k(345.5)420.0 k(414.5)560.0 k(518.2)560.0 k(552.7)640.0 k(621.8)720.0 k(690.9)900.0 k(863.6)1200 k(1036)
1400 k(1382)
1800 k(1727)
2400 k(2073)
2800 k(2591)
2800 k(2764)
3200 k(3109)
3600 k(3455)
4200 k(4145)
5600 k(5182)
7200 k(6909)
12.00 M
(10.36)150V(110V)
[W]

1000

500.0

1200

600.0

1500

750.0

1600

800.0

2000

1000

2400

1200

3000

1500

4000

2000

Multiplying
factor

×0.01

5000

2500

6000

3000

8000

4000

10.00 k5000

12.00 k6000

15.00 k7500

16.00 k8000

20.00 k10.00 k24.00 k12.00 k30.00 k15.00 k40.00 k20.00 k50.00 k25.00 k60.00 k30.00 k80.00 k40.00 k100.0 k50.00 k120.0 k60.00 k150.0 k75.00 k160.0 k80.00 k180.0 k90.00 k200.0 k100.0 k240.0 k120.0 k300.0 k150.0 k320.0 k160.0 k360.0 k180.0 k400.0 k200.0 k500.0 k250.0 k600.0 k300.0 k800.0 k400.0 k1000 k

500.0 k1200 k

600.0 k1500 k

750.0 k1600 k

800.0 k1800 k

900.0 k2000 k1000 k2400 k1200 k

3000 k1500 k4000 k2000 k6000 k

3000 k

×0.1

×1

×10

×100

<Note 1>
Parenthesis is primary active power (reactive power) value in /500W(500var).
In the blank, setting is impossible. A display of an default is 4 digits.
The unit may be changed if the number of digits is changed.
An example) 4 digits
3 digits
4800W→ 4.80kW
4000kvar → 4.00Mvar
2000kW → 2.00MW

68<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a 4-step setup
is possible) is set as the fastest, theoutput pulse width of a pulse output is set
to 100 to 130ms. (Usually 240 to 260ms)

SFLC-212-173

Tokyo Office : 11-13, Hitotsuya 1-chome, Adachi-ku, Tokyo, 121-8639, JAPAN.
TEL：+81-3-3885-2411 , Fax：+81-3-3858-3966
Kyoto Office : 1-19, Ichinobe-Nishikawahara, Jyoyou-shi, Kyoto, 610-0114, JAPAN.
TEL：+81-774-55-1391 , Fax：+81-774-54-1353
Revision C, DATE：February 20, 2023

69