― 計測システムの総合メーカ ―Digital Power Multi Meter
Digital 3 Phase
Ammeter & Voltmeter
SFLC-110L・SALC-110L・SVLC-110L

Lineup a New English Specification U-V-W Display for 3 Type Digital Meter

ＳＦＬＣ－１１０Ｌ(110×110×104mm) / 600g

ＳＡＬＣ－１１０Ｌ

ＳＶＬＣ－１１０Ｌ(110×110×104mm) / 600g

(110×110×104mm) / 600g

Digital Power Multi Meter / Digital 3 Phase Ammeter & Voltmeter

SFLC-110L / SALC-110L / SVLC-110L

Lineup a New English Specification U-V-W Display together with Japanese Specification R-S-T Display &English Specification R-Y-B Display. English Specification will display the guidance in English.
Ex.: 需要 (Japanese) → DEMAND (English)

Digital Power Multi Meter ＳＦＬＣ－１１０Ｌ■Features
3 Voltage, 3 Current, 3 Demand Current, Power, Demand Power, Reactive Power, Power Factor, Frequency, Active Power, Reactive Energy
can be measurement by 1-unit Digital Power Multi Meter. And also can monitoring incoming circuit, save energy for power, demand current form
low voltage to high voltage. Option for digital input, alarm and pulse output also can add on together with system centralized monitoring.

■Type Name & Specification Code
①Type NameSFLC-110L

Specification Code－③ InputCircuit② Hard Model
R-S-T Display
Japanese
R-Y-B Display
EEnglish
U-V-W Display
F
English④ InputRange⑤ Power Supply ⑥ Digital Input⑦ Analog Output OR－ CommunicationOutput

0 None

DWhiteBacklight

1P2W
1P3W
3P3W

F150V
300V

F

AC85～264V
1 DC80～143V
Both use⑨ View
Position⑧ Digital Output

0 NoneFor Hard Model
D use only

0 None1 4～20mA×3
1 1 Circuit

F Wide View

5A

For Hard Model
E & F use

1 Pulse & Alarm

A Protocol A
M Modbus RTU

1 Upper View

2 Lower View
Note: Only 4 combination code for this SFLC-110L Type
Code「0-00」「1-11」「1-A1」「1-M1」 for Digital Input, Analog Output & Digital Output
Please contact us for more detail specification

Digital 3 Phase Ammeter ＳＡＬＣ－１１０Ｌ■Features
3 Current, 3 Demand Current can be measurement by 1-unit Digital 3 Phase Ammeter. This is most suitable for load monitoring transformer,feeder etc. by measurement current incoming circuit. Option like Analog output, Alarm output and Digital Input can add on together with systemcentralized monitoring.

■Type Name & Specification Code
①Type NameSALC-110L

Specification Code－③ Inputsircuit② Hard Model
R-S-T DisplayJapanese
R-Y-B Display
EEnglish
U-V-W Display
F
English④ InputRange⑤ Power Supply

D1WhiteBacklight
F

1P2W
1P3W
3P3W

55A

AC85～264V
DC80～143V
Both use⑥ DigitalInput⑨ View
Position－ ⑦ Analog Output ⑧ Digital Output

0 None0 None0 NoneFor Hard Model
D use only

1 1 Circuit

1 4～20mA×31 Alram

F Wide View

Note: Only 2 combination code for this SALC-110L TypeFor Hard Model
E & F use

Code「0-00」「1-11」for Digital Input, Analog Output & Digital Output

1 Upper View

Please contact us for detail specification

2 Lower View

Digital 3 Phase Voltmeter ＳＶＬＣ－１１０Ｌ■Features
3 Voltage and Frequency can be measurement by 1-unit Digital 3 Phase Voltmeter. This is most suitable for measurement voltage atincoming, bus side and voltage system. Option like Analog output, Alarm output and Digital input can add on together with system centralized
monitoring.

■Type Name & Specification Code
①Type Name
SVLC-110L

Specification Code－③ Input
Circuit② Hard Model
R-S-T Display
JapanesWhite
R-Y-B Display
E
EnglishBacklight
U-V-W Display
F
English④ InputRange⑤ Power Supply

D1F

1P2W1P3W
3P3W

D150V
300V

AC85～264V
DC80～143V⑥ Digital Input－ ⑦ Analog Output⑧ Digital Output⑨ View
Position0 None0 None0 NoneFor Hard Model
D use only

1 1 Circuit

1 4～20mA×31 Alarm

F Wide View

Note: Only 2 combination code for this SVLC-110L TypeFor Hard Model
E & F use

Code「0-00」「1-11」for Digital Input, Analog Output & Digital Output

1 Upper View

Please contact us for more detail specification

2 Lower View

http://www.daiichi-ele.co.jp

IP1304-1