Complex ExtraSoft • Soft
MediumSoft • Standard
Complex

Framing that is stable & easily fits into aneurysms●

Designed to sit in open space
The distal coil shape spreads along the aneurysm wall & the proximal
coil shape provides support from the inside.

●

Coil designs optimized according to size
Combination of Primary Coil Diameter & length for easy delivery &stability.

Complex SilkySoft
ComplexPursuit of softness, SilkySoft●

3D & Helical Hybrid
Straight, in-line structure coil design searches for space while entwinedwith other coils.

●Softest coil in our product historySoft coils & the Waving SR makes tight packing & minimal catheter
kick-back possible.

Refined Pushability with
Unique Gold-colored
Non-Slip Coating

Complex ∞SilkySoft

Complex Inﬁni

New concept filling coils●12 Coil Type SilkySoft
A soft coil with a 0.012” Primary Coil Diameter for volume maximization.

●

Filling coil that moves into space with ease
Large loops that can fold tightly, configure according to the size of thespace and pack them.

●Greater volume gain performance with a single coil
Due to the Waving SR and softness of the SilkySoft, the coil can foldcompactly enabling choice of longer coils possible.

14∞Inﬁni

ExtraSoft • Soft • Standard

New concept multi-utility coils●Low Shape-Memory
Outward behavior coils actively seek open space, fitting into various size& shape aneurysms.
Possible to select longer length coils, regardless of Secondary CoilDiameter, enabling volume maximization with fewer coils.

●Utility player coil in various situations
Applicable in Stent Assist / Vessel Occlusions / Dissecting Aneurysm /dAVF / Sinus Packing.

●Easy choice of coils in use with stents
Coil selection by length & softness.

Sound & Light System To Signal Best Detach Point

EDGv4 indicates by Sound and Light
ght to signal the best Detach Point even when
position of 2nd marker of Microcatheter
theter is not detectable under X-Ray Fluoroscopy.

Product Specifications
Pusher Length 187cm

Type

Catalogue No.

Primary CoilDiameterinch

Secondary CoilDiameter*mm

Primary Coil Lengthcm
1390-0101
390-0102

12390-0103

3390-01H22

1.5

390-01H3

3390-0203

SilkySoft(SS)

390-0204
390-020630.010426390-0208

8390-02H4

42.5

390-02H6

6390-0304

4390-0306

36390-0308

ExtraSoft(ES)

8392-0304
392-03H80.010

392-0408
392-0510

MediumSoft(MS)
Standard(STD)

Type392-0515
392-0620

63.5

884392-0410

Soft(S)43392-0306

101050.01215

620

392-0730

730

392-0840

840

392-0940

940392-1040

1040392-1250

0.014

1250

392-1450

1450

392-1650

1650

Catalogue No.

Primary CoilDiameterinch

Secondary CoilDiameter*mm

Primary Coil Lengthcm
4393-020304

SilkySoft
(SS)

393-020306
393-020310

2-3393-030515

Soft(S)
Standard(STD)

10153-5393-030520

ExtraSoft
(ES)

60.012

2010373-040810
373-040815

4-815373-040830
373-051230

300.014373-051250
373-122050

305-1250

12-2050＊ The secondary coil diameter means referenced diameter of a space to be embolized.

TypeCatalogue No.

Description

EDG v4347-404

ELECTRO DETACH GENERATOR v4

Manufacturer
3-18, 2-Chome, Nakanoshima, Kita-ku, Osaka-city, OSAKA, 530-8288 JAPAN
TEL.+81-6-6226-5256
https://www.kaneka-med.jp/english/

Please scan QR Code to contact

Ref No.107-245 Date Oct.2021 2000 DI