Tierrett Nelaton Catheter for Dogs
Instructions for use
Contraindicated to re-use[Precautions][CONTRAINDICATIONS・PROHIBITION]
Do not reuse the product (single use only).

Carefully apply to the following patients;[1] Confirm in advance if it is applicable to cases with urethral stenosis.
[There is a risk of damage to tissues and urethral mucosa.][Shape, structure, principle]
This product is sterilized with ethylene oxide gas.

<Important basic caution><Shape>[1] Manage appropriately the catheter state during placement. Check theplacement state as required.
[The catheter lumen may be occluded due to breakage, bending, twisting,

Catheter connector

urine components, stones, etc. of the catheter.][There is a risk of ulceration, perforation, etc. when the tip of the catheter

Catheter tubecomes in contact with the bladder wall.][2] Do not pinch the device with forceps too strongly.
Ｃ[The tube may be broken or lumen may be occluded.]Ｂ[3] Note that di (2-ethylhexyl) phthalate, which is a plasticizer for polyvinyl
chloride, may elute in fat-soluble pharmaceuticals or chemicals.
[Polychlorinated vinyl chloride is used for the connection 2 port adapter.]

ＡA:

B:

C:

Total length

Effective length

O.D.

8Fr

400mm

370mm

2.7mm

10Fr

400mm

370mm

3.3mm

8Fr (Long type)

580mm

550mm

2.7mm

constituents or blood clots etc.][2] Cut of the catheter.

10Fr (Long type)

580mm

550mm

3.3mm

[Cut due to the following causes]

Size<Failures >[1] Occlusion of the catheter.
[The lumen of the catheter may be occluded by the adhesion of the urinary

・Damage caused by forceps, scissors, knife or other apparatuses<Raw Materials>

・Damage due to the calculus in the patient.

Catheter: PVC (polyvinyl chloride)・Sudden load on the product such as self (accidental) removal.
・Other combined causes due to the above events.

<Principles>
Insert the catheter into the bladder. Urine is discharged from the side hole throughAdverse eventsthe lumen and through the urine outlet.

The following adverse events may be caused by the use of the product:
・Urethral infection, bloodstream, fever, pain・Urethral damage, ulceration, perforation, hematuria (bleeding), bladder stones[Intended purpose, efficacy or effect]
Placed in the bladder and used for urinary catheterization.

・Urine leakage from the side of the catheter, urinary incontinence after removal
・Urethral edema and ulcer around the catheter[Operating or using method]
The below is a general procedure.

・Remains in the body due to catheter cutting[1] Disinfect (clean, wipe) the external urethral meatus and vulva as required.

[Storage conditions and duration of use]
<Storage conditions>Store the product hygienically, avoiding the direct sun light, high humidity andultraviolet rays such as a sterilizing lamp and taking care of wetting.

[2] Remove the product from the packaging material hygienically and insert into
the urethra and bladder.
[3] Confirm that the tip of the catheter reaches the inside of the bladder (passes
through the external urethral sphincter muscle) and urine begins to flow out< Expiration date >from the urine outlet

See the expiration date given on each package provided that the device is stored
appropriately.

[4] After confirming the drainage, advance about 2 cm further and connect theurine outlet of the catheter to a urine bag etc. as required.

[By self-authentication (our data).][5] When removing the catheter, remove the urine bag etc. connected to thecatheter, after release the catheter, pull out slowly and gently.

Manufacturer<Precautions for the using method>[1] When connecting a urine bag etc. to the catheter urine outlet, make sure toCREATE MEDIC CO.,LTD.
2-5-25 Chigasakiminami, Tsuzuki-Ku, Yokohama, Japan

select one fits. Check for leaks and looseness at the connections during use,make sure to use it in a connected state.

In-house administration number: DC67932

1／1

R0-0 2020.10.29

ＤＣ６７９３２