Display Color Analyzer CA-410
CIE 170-2:2015 Compliant Probe

CA-P427C/P410C
CIE 170-2: 2015 is a chromaticity diagram with physiological axes definedby CIE and announced in 2015. This probe has color-matching functions
conforming to the CIE 170-2: 2015 definition, and provides luminance andchromaticity data that have high correlation with visual results even when
adjusting wide color gamut displays.
CA-P427C

1.0- （λ）
ｘ- （λ）
yCIE1931
color-matchingz- （λ） function

Relative spectral sensitivity

0.9- （λ）
ｘ- （λ）
yCA-P410CModelMeasurement area
Acceptance angle
Accuracy guaranteed measurement distanceLuminance
Display range
Chromaticity
Accuracy guaranteed range

Luminance

Chromaticity

Flicker(Contrast)> 0.001 cd/m2> 0.01 cd/m2> 0.1 cd/m2
Accuracy (for white)*1, *3> 1 cd/m2> 10 cd/m2> 100 cd/m2> 0.001 cd/m2> 0.01 cd/m2> 0.1 cd/m2
Repeatability AUTO
*1(2σ)> 1 cd/m2> 10 cd/m2> 100 cd/m2
Accuracy guaranteed luminance range> 0.01 cd/m2> 0.1 cd/m2*1, *3
Accuracy (for white)> 1 cd/m2> 10 cd/m2> 100 cd/m2
At 100 cd/m2 (for100cd/m2monochrome) *2> 0.01 cd/m2> 0.1 cd/m2
Repeatability AUTO
> 1 cd/m2*1(2σ)> 10 cd/m2> 100 cd/m2
Measurement luminance range
Measurement target (Flicker frequency)
30 Hz, AC/DC
10% sine wave
Accuracy
60 Hz, AC/DC
10% sine wave
20-65 Hz, AC/
Repeatability (2σ)
DC 10% sinewave

CA-P427C

Relative spectral sensitivity

Normal probe
CA-P427z- （λ） spectral responsivity

1.00.90.8
0.7
0.60.50.4
0.3
0.2
0.1
0.0
380

0.8
0.7

430

480

530580

630

680

730

CA-P410C

Ф 10 mm± 5°
30 ± 5 mm
0.0001 to 5,000 cd/m2
Displayed in 4 digits
0.01 to 5,000 cd/m2--± 2.5 %±2%±2%± 1.5 %
± 1.5 %--2%0.60 %
0.20 %
0.10 %
0.10 %
0.01 to 5,000 cd/m2± 0.006± 0.002± 0.002± 0.002± 0.002± 0.003± 0.003

0.00350.00150.00040.00030.0002
5 to 1,500 cd/m2
0.25 to 65 Hz

0.00700.00200.00080.00050.000315 to 3,000 cd/m2
0.25 to 65 Hz± 0.4%± 0.4%

0.3%

0.3%

- （λ）
ｘF- （λ）
yCA-P427C
spectral
responsivity

Fz-F（λ）

0.6
0.5
0.4
0.3
0.20.1

Ф 27 mm± 2.5°
30 ± 10 mm
0.0001 to 5,000 cd/m2
Displayed in 4 digits
0.001 to 5,000 cd/m2±9%±2%± 1.5 %
± 1.5 %
± 1.5 %
± 1.5 %
10 %1%
0.40 %
0.10 %
0.10 %
0.10 %
0.01 to 5,000 cd/m2± 0.003± 0.002± 0.002± 0.002± 0.002± 0.7%

CIE170-2:2015
color-matching
functionFz-F（λ）

780

0.0
380

430

480

Flicker(JEITA)

Accuracy
guaranteed
measurement
speed
*4

530

580

630

680

730

780

Wavelength (nm)Wavelength (nm)± 0.7%- （λ）
ｘF- （λ）y

Measurement luminance range
Measurement target (Flicker frequency)
30 Hz, AC/DC
4% sine wave
Accuracy
30 Hz, AC/DC
1.2% sine wave
30 Hz, AC/DC
4% sine wave
Repeatability (2σ)
30 Hz, AC/DC
1.2% sine wave
Lvxy

AUTO

Flicker (Contrast)
Flicker (JEITA)

Measurement synchronization mode
Measurement speed mode
Measurement target (Vertical synchronization frequency)
User calibration memory channel
Communication
Interface
TriggerSize (mm)WeightPower supply
Operation temperature/humidity range*5
Storage temperature/humidity range
StandardAccessories
Optional

5 to 1,500 cd/m2
0.42 to 65 Hz

15 to 3,000 cd/m2
0.42 to 65 Hz± 0.35dB± 0.35dB± 0.35dB± 0.35dB

0.1dB

0.1dB

0.3dB

0.3dB

1 time/sec (> 0.001 cd/m2) 1 time/sec (> 0.01 cd/m2)
5 times/sec (> 0.15 cd/m2) 5 times/sec (> 0.15 cd/m2)
20 times/sec (> 2 cd/m2)
20 times/sec (> 2 cd/m2)
20 times/sec
20 times/sec
0.5 times/sec (at 1HzPitch), 0.5 times/sec (at 1HzPitch),
2.5 times/sec (at 10 HzPitch) 2.5 times/sec (at 10 HzPitch)
NTSC, PAL, EXT, UNIV,NTSC, PAL, EXT, UNIV,
INT, MANU (4ms to 4s)
INT, MANU (4ms to 4s)
INT, MANU (4ms to 4s)
INT, MANU (4ms to 4s)
AUTO, LTD. AUTO,
AUTO, LTD. AUTO,SLOW, FAST
SLOW, FAST
0.5 to 240 Hz (luminance 0.5 to 240 Hz (luminanceand chromaticity) 0.5 toand chromaticity) 0.5 to
130 Hz (flicker)
130 Hz (flicker)
99 channels
99 channels
USB2.0, RS-232C
USB2.0, RS-232C
In ＆ Out [5V]
In ＆ Out [5V]
42 x 42 x 139.7
42 x 42 x 173.5
270 g (including
280 g (includingmount)
mount)
DC 5 V (input from USB bus power line or RS
communication connector)
10 to 35°C, relative humidity 85% or less with nocondensation
0 to 45°C, relative humidity 85% or less (at 35°C)
with no condensation
PC Software for Color Analyzer Ver.1.0 CA-S40*6,
SDK for Color Analyzer CA-SDK2, USB Cable forProbe-PC (2 m) IF-A28, Hood for Probe, Lens
Cap for Probe
Conversion Cable IF-A29, BNC Conversion CableIF-A35*1: Measured under Konica Minolta’s standard light source (6,500K).
*2: Luminance for monochrome is measured when reading of luminance for white is 100 cd/m2.
*3: Temperature 23°C/±2°C, relative humidity 40%±10%*4: In NTSC synchronization mode using USB with one probe*5: Reading fluctuation (compared to reference reading at 23°C, 40% RH): Luminance: ±2% for white; Chromaticity (at100 cd/m2): ±0.002 for white, ±0.003 for monochrome*6: The probe corresponding to CIE 170-2: 2015 can not be used under Ver. 1.30 or earlier.

Merits of using a CIE 170-2:2015-compliant probe
When adjusting color of conventional displays using our color analyzer probe conforming to CIE 1931
Conventional display

No visual difference
was detected between
conventional displays.
White

y0.90
0.80
0.70
0.60
0.50
0.40

Wide color gamut display

0.30
0.20
0.10
0.00

0.10 0.20 0.30 0.40 0.50 0.60 0.70

xLv ＊＊＊
ｘ ＊＊＊＊
ｙ ＊＊＊＊WHITE!
Slightly
yellowish
white!

Differences are visuallyevident even though measured
values are matched to displaysof wide color gamut.

Issue

Because of the advanced color technologies used with
OLED, quantum dot, laser and other advanced displays,
colors need to be adjusted between displays of differing
spectral characteristics. However, even when measured
values are matched, colors do not always look the same.

Improvements made using CIE 170-2:2015
When adjusting color of wide color gamut displays using our CIE 170-2:2015-complaint probe
Wide color gamut display
y0.90
0.80
0.700.60
0.50
0.40
0.30
0.20
0.10
0.00

0.10 0.20 0.30 0.40 0.50 0.60 0.70

xLv ＊＊＊
ｘ ＊＊＊＊
ｙ ＊＊＊＊WHITE!

WhiteColors can be adjusted to look the same even with displays of wide color gamut.
KONICA MINOLTA, the Konica Minolta logo and symbol mark, and “Giving Shape toideas” are registered trademarks or trademarks of KONICA MINOLTA, INC.
Screens shown are for illustration purpose only.
The specifications and appearance shown herein are subject to change without notice.
Some lighting control methods may make accurate measurements difficult. For details,
please contact your nearest Konica Minolta sales office or dealer.

Dimensions (Units : mm)
CA-P427C

CA-P410C
ø26.5
ø20.8

13612611628138.2

73.7

421.5

ISO 5mmDepth 6mm
Tripod socket Depth 6mm
ISO 5mmDepth 6mm

Tripodsocket
Depth6mm
11When usedwith the hood

ISO 5mmDepth 6mm
ISO 5mmDepth 6mm

32107.6

ø33

When usedwith the hood108988828

ø37

Tripod socket
Depth 6mm
Tripod socket
Depth 6mm
1142

ISO Certifications of KONICA MINOLTA, Inc., Sakai Site

SAFETY PRECAUTIONS
For correct use and for your safety, be sure to read the instruction manual before using the instrument.
Always connect the instrument to the specified power supply voltage. Improper connection may cause a fire orelectric shock.

©2018 KONICA MINOLTA, INC.

JQA-QMA15888
Design, development, manufacture/manufacturing management, calibration, andservice of measuring instruments

JQA-E-80027
Design, development,
manufacture, service and salesof measuring instruments

9242-4885-20 BJDPK 3