Product Information

PI"SD Series I.I."
X-ray Image Intensifier

E5764SD-P7A
Triple Input Field:
230/160/120mm (9”/ 6”/ 4.5”)
Output Image Diameter: 20mm

FEATURING: High Resolution: 48 Lp/cm
cd/m2
 High Conversion Factor : 28 μGy/s
 High Contrast Ratio
10% area contrast: 25:110mm dia. contrast: 16:1“SD Series I.I.” E5764SD-P7A is used for medical radiographic diagnosis.

This X-ray Image Intensifier is intended to be connected with IEC ClassⅠequipment.

“SD Series I.I.” E5764SD-P7A has been perfected by the integration of several aspects of advanced
technology；
The input window is of a thin metal plate, which has good X-ray transparency characteristics andreduces X-ray scatter.
It uses a thick input phosphor screen made with an extremely fine pillar shaped structure. This has
high X-ray absorption and very high conversion efficiency. In addition, there is a thin output
phosphor screen on a single thick-glass output window with an anti-reflective coating, offering very
high contrast for small details, very low structure noise and high MTF characteristics.
The E5764SD-P7A, employing these manufacturing techniques, provides outstanding performancein resolution, brightness, contrast and noise.
The E5764SD-P7A is supplied in a housing, which provides mu-metal magnetic shielding andradiation shielding.

★The information contained herein is presented only as a guide for the application of our products. No responsibility is assumed by Canon Electron Tubes& Devices Co., Ltd. (CETD) for any infringements of patents or other rights of the third parties which may result from its use.
No license is granted by implication or otherwise under any patent or patent rights of CETD or others.
★The information contained herein may be changed without prior notice. It is therefore, advisable to contact to CETD before processing with the design ofequipment incorporating this product.

E5764SD-P7A

PERFORMANCE DATA
Item
Central Resolution
Conversion Factor (Gx)
10% area
10 mm dia.
DQE (IEC standard)
Contrast Ratio

N mode
48
28
240
25:1
16:1
65

M1 mode56-

M2 mode66-

Lp/cm(cd/m2)/(μGy/s)(cd/m2)/(mR/s)%GENERAL DATA MECHANICAL:Item
Overall Length
Maximum Diameter
Nominal Entrance Field Size
N (9″) mode
Useful Entrance Field Size
N (9″) modeM1 (6″) modeM2 (4.5″) mode
Output Image Diameter
Weight (approx.)
Dimensional Outline

341±5 mm
302±2 mm
230 mm min.
215 mm min.
160±5 mm
120±5 mm
20±0.5 mm
18 kg
Refer to page 5

 POWER SUPPLY:
ItemHigh Voltage Power Supply
Input Voltage

24±1 Vdc

 ENVIRONMENTAL:
Item
Temperature
Humidity
Pressure

Under delivery and stock-15 to 45 ℃
10 to 90 %50 to 106 kPa-2-Under operating
5 to 40 ℃
30 to 85 %70 to 106 kPatyp.
typ.
typ.
typ.
typ.
typ.

E5764SD-P7A

ABSOLUTE MAXIMUM RATINGS
ItemAnode Voltage
G3 Electrode Voltage
Potential Difference between Anode andG3 Electrode
Potential Difference between G3 Electrodeand G2 ElectrodeG2 Electrode Voltage
G1 Electrode Voltage
Photocathode Current
ContinuousPulsed
Input X-ray Dose

32 kV
12 kV
30 kV
12 kV
1200 V
250 V
0.2 μA
2 μA
8.73×10-3Gy/min [1R/min]-3-

E5764SD-P7A

ELECTRICAL WIRING AND CONNECTING
X-ray image intensifier is operated with 24Vdc input voltage applied to the high voltage power supply.
Wiring and connecting are shown as follows.
DIN Connector

1Pin No.
12345Signal
DC24V input
NC0V
M1M2

Switching of entrance field size
* N mode4, 5 Open* M1 mode
1-4 Close (apply DC24V to Pin No.4)* M2 mode
1-5 Close (apply DC24V to Pin No.5)
1. DIN connector ...................................................................................................................... 1 pc-4-

E5764SD-P7A

DIMENSIONAL OUTLINE
Unit: mmⒶⒷⒸⒶ,Ⓑ,Ⓒ: Screw holes for fixing to apparatus-5-

E5764SD-P7A

Marketing Engineering Group, Sales Department
1385, Shimoishigami, Otawara-shi, Tochigi 324-8550, Japan
Tel: +81-287-26-6666
Fax: +81-287-26-6060
https://etd.canon･The head office of Canon Electron Tubes & Devices Co., Ltd. has been certified to meet all the requirements of Environmental Management System ISO14001.
･Canon Electron Tubes & Devices Co., Ltd. has been certified to meet all the requirements of Quality Management Systems ISO9001 and ISO13485.
Product scope is referred to the following URL. https://etd.canon/eng/company/quality.htm.

PE-E5764SD-P7A 2018-11