BA00413P/00/EN/13.16
71335863

Products

Solutions

Services

Valid for software version:
02.10.54

Operating Instructions – Descriptionof Device Functions
Cerabar S PMP71
with MID Part Certificate
Process pressure measurement

Cerabar S 4 to 20mA HART

1.

Order code: XXXXX-XXXXXX
Ser. no.:XXXXXXXXXXXX
Ext. ord. cd.: XXX.XXXX.XX

Serial number

2. www.endress.com/deviceviewer

Endress+Hauser
Operations App

3.

A0023555

Make sure the document is stored in a safe place such that it is always available whenworking on or with the device.
To avoid danger to individuals or the facility, read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures.
The manufacturer reserves the right to modify technical data without prior notice. Your
Endress+Hauser Sales Center will supply you with current information and updates to these
Instructions.

2

Endress+Hauser

Cerabar S 4 to 20mA HART

Table of contents
1

Notes on use . . . . . . . . . . . . . . . . . . . . . . . 4

1.1
1.2

Finding parameter description using ID numbers . 4
Finding function group using graphic
representation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Finding parameter description using parameternames (index) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

1.3

2

Finding parameter description using ID
numbers. . . . . . . . . . . . . . . . . . . . . . . . . . . 5

3Graphic representation of functiongroups . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

4Pressure measurement . . . . . . . . . . . . . . 8

4.1
4.2

Calibration with reference pressure . . . . . . . . . . . . 8
Calibration without reference pressure . . . . . . . . . 95Description of parameters . . . . . . . . . . 11

6

Troubleshooting . . . . . . . . . . . . . . . . . . . 39

6.1
6.2
6.3

Messages . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Response of outputs to errors . . . . . . . . . . . . . . . 46
Confirming messages . . . . . . . . . . . . . . . . . . . . . . 47

7

Appendix . . . . . . . . . . . . . . . . . . . . . . . . . 48

7.1

Operating menu for onsite display and digitalcommunication . . . . . . . . . . . . . . . . . . . . . . . . . . . 48

Index. . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55

Endress+Hauser

3Notes on use

Cerabar S 4 to 20mA HART

1

Notes on use

Section 5 describes all the parameters in the order of how they appear in the menu. Section 4
describe typical configuration examples.
Section 1.1 to Section 1.3 describe ways of finding a certain parameter description moreeasily.

1.1

Finding parameter description using ID numbers

Each parameter is shown on the onsite display with a unique identification number (ID). All
the parameters are listed in numerical order in Section 2. The page reference/link takes you
to the parameter in question.
In the operating program, additional parameters and, to an extent, other parameters are
displayed. These parameters are not shown in Section 2. You can find these parameters bymeans of the index. See also Section 1.3.

1.2

Finding function group using graphic representation

Section 3 provides an outline of all of the function groups in tables. The page reference/linktakes you to the function group in question. In Section 5, all of the parameters in a functiongroup are summarized in a table.

1.3

Finding parameter description using parameternames (index)

The index lists all the parameters in alphabetical order. The page reference/link takes you
to the parameter in question.

4

Endress+Hauser

Cerabar S 4 to 20mA HART

Finding parameter description using ID numbers

2

Endress+Hauser

Finding parameter description using ID
numbers

ID number

Parameter name

Description, see Page

014
015
016
017
018
021
022
036
042
043
044
045
046
047
048
055
060
075
079
245
246
247
250
254
264
266
270
271
272
301
305
309
310
317
318
319
332
333
334
335
336
339
342
343
345
350
352
354
357
358
359
360
362
363
365
366
367
368
369
378
386
388

DOWNLOAD SELECT
FULL PRESSURE
EMPTY PRESSURE
FULL CALIB.
EMPTY CALIB.
SET LRVSET URVPREAMBLE NUMBER
CURR. TRIM 20mA
OFFSET 4mA TRIMOFFSET 20mA TRIM
CURR. TRIM 4mA
ALARM STATUS
ENTER RESET CODE
INSERT PIN No
CUST. TAG NUMBER
PRESS. ENG. UNIT
CUSTOMER UNIT P
LANGUAGE
SET LRV – "Pressure" measuring mode
SET URV – "Pressure" measuring mode
DAMPING TIMESENSOR SER. No.
OUTPUT CURRENTSOFTWARE VERSIONHARDWARE REV.
SIM. CURRENT
HART MESSAGE
ADDITIONAL INFO.
PRESSURE – "Pressure" measuring mode
LONG TAG NUMBER
GET LRVGET URV
CUST.UNIT FACT.P
TEMP. ENG. UNIT – "Pressure" measuring mode
CALIB. OFFSET
Pmin ALARM WINDOW
Pmax ALARM WINDOW
Tmin ALARM WINDOW
Tmax ALARM WINDOWALARM DELAYDISPLAY CONTRAST
SET MAX. ALARM
SET MIN. CURRENTBUS ADDRESSDEVICE DESIGN.
CONFIG RECORDER
DEVICE SERIAL No
PCB TEMPERATURE
ALLOWED MIN. TEMPALLOWED MAX. TEMP
MAT. PROC. CONN. +SEAL TYPEDIP STATUS
MAT.
FILLING FLUIDSENSOR TEMP.
Tmin SENSOR
Tmax SENSOR
MEAS. VAL. TRENDELECTR. SERIAL NO.
OUTPUT FAIL MODE33
See 1)
See 1)
See 1)
See 1)
See 1)
See 1)
25
38
38
38
38
35
32
33
26
16
16
11
13 or 16
13 or 16
13, 17
29
21
26
26
34
25
26
31
26
17
17
16
18
14
36
36
37
37
36
20
23
23
24
26
26
26
26
26
26
27
28
27
29
29
31
29
29
31
26
22

5

Finding parameter description using ID numbers

ID number

Parameter name

Description, see Page

389
401
409
413
414
419
423
432
434
476
480
481
482
484
485
487
500
563
564
570
581
584
591
595
597
600
603
679
688
694
696
699
764
802
831
832
836
838
840
841
844
845
847
852
853
855
856
875

MEASURING MODE
ACK. ALARM MODE
OPERATING HOURSSIMULATION MODE
SIM. PRESSURE
MENU DESCRIPTOR
ALTERNATE DATAMANUFACTOR ID
CORRECTED PRESS. – "Pressure" measuring mode
SIM. ERROR NO.
ALARM DISPL. TIME
HART DATEPROC. CONN. TYPE
PRESS. SENS LOLIMPRESS. SENS HILIMSENS H/WARE REV
ACK. ALARM
POS. INPUT VALUELAST DIAG. CODE
Pmax PROC. CONN.
SENSOR MEAS.TYPE
SENSOR PRESSURE – "Pressure" measuring mode
MINIMUM SPAN
SELECT ALARMTYPE
ALT.CURR.OUTPUTSELECT ALARMTYPERESET ALL ALARMS
MEASURED VALUE – "Pressure"
MAIN DATA FORMATCURR. CHARACT. – "Pressure"
CURR. CHARACT. – "Height"
DEVICE REVISION
CURR. CHARACT. – "Tank content"
DEVICE TYPE, Cerabar S
HistoROM AVAIL.
HistoROM CONTROLSAFETY LOCKSTATESAFETY PASSWORD
DIGITS SETDIGITS SETACK. ALARM MODE
MEASURING MODE
CALIB. OFFSET
SET LRVSET URVDAMPING TIME
CONF. PASSWORD
CURRENT OUTPUT

12, 15
35
32
34
34
19
19
25
31
34
36
25
27
29
29
29
35
13 or 14
35
27
29
31
29
36
22
36
35
30
19
22
22
24
22
24
33
33
See 1)
See 1)
20
See 1)
See 1)
See 1)
See 1)
See 1)
See 1)
See 1)
See 1)
See 1)

1)

6

Cerabar S 4 to 20mA HART

For further information, see the Cerabar S Functional Safety Manual (SD00190P).

Endress+Hauser

Cerabar S 4 to 20mA HART

Graphic representation of function groups

3

Graphic representation of function groups

1st Selection level

2nd Selection level (groups)

Function groups

LANGUAGELANGUAGE (079)

11

OPERATING MODE

OPERATING MODE (389)



11



12

 POSITION ADJUSTMENT

14

 BASIC SETUP pressure15

 EXTENDED SETUP pressure

18

 SAFETY CONFIRM.



See 1) DISPLAY (558)19 OUTPUT (559)21 HART DATA24

 TRANSMITTER DATA26

 PROCESS CONNECTION27

 SENSOR DATA29

 PROCESS VALUES pressure



30



32

 SIMULATION MODE34

 MESSAGES35

 USER LIMITS36

 SYSTEM 238

QUICK SETUP pressure
OPERATING MENU (555) SETTINGS (557) TRANSMITTER INFO (560) PROCESSINFO (561)
 OPERATING DIAGNOSTICS (562) SERVICE (561)1)

Description,
see Page

For further information, see the Cerabar S Functional Safety Manual (SD00190P).

Endress+Hauser

7

Pressure measurement

Cerabar S 4 to 20mA HART

4

Pressure measurement

4.1

Calibration with reference pressure

Example:In this example, a device with a 10 bar (150 psi) sensor is configured for the 0 to +4 bar(60 psi) measuring range, i.e. 0 bar is assigned to the 4 mA value and 4 bar (60 psi) to the
20 mA value.
Prerequisite:• The pressure values 0 bar and 4 bar (60 psi) can be specified. The device is alreadyinstalled, for example.
• See also Operating Instructions for Cerabar S (BA00412P), "Pressure measurement"
section.
• For a description of the parameters mentioned, see– Page 11, Table 1: MEASURING MODE
– Page 14, Table 3: POSITION ADJUSTMENT– Page 15, Table 4: BASIC SETUP.
• For a description of further relevant parameters, see– Page 18, Table 5: EXTENDED SETUP– Page 30, Table 12: PROCESS VALUES.

8

Endress+Hauser

Cerabar S 4 to 20mA HART

Pressure measurement

Description
1

Carry out position adjustment if necessary. See Page
14, Table 5: POSITION ADJUSTMENT.

2

If necessary, select the "Pressure" measuring mode
via the MEASURING MODE parameter.
!

I
[mA]

➁ 20

WARNING

Changing the measuring mode affects the span
(URV)!
This situation can result in product overflow.
If the measuring mode is changed, the spansetting (URV) must be verified in the"CALIBRATION"  "BASIC SETUP" operating
menu and, if necessary, reconfigured!

‣➀ 4

Onsite display:
Menu path: GROUP SELECTION MEASURING
MODE
Digital communication
Menu path: OPERATING MENU SETTINGSBASIC SETUP MEASURING MODE
3

Onsite display:
Select BASIC SETUP function group. Menu path:
GROUP SELECTION OPERATING MENU 
SETTINGS BASIC SETUP

4

Select a pressure unit via the PRESS. ENG. UNIT
parameter, here mbar for example.

5

The pressure for the lower range value (4 mA value)is present at the device, here 0 mbar for example.

0

4

p
[bar]

P01-MIDxxxxx-05-xx-xx-xx-004

Fig. 1:
1
2

Calibration with reference pressure
See Table, Step 5.
See Table, Step 6.

Select GET LRV parameter.
Confirm value present. The pressure value present is
assigned to the lower current value (4 mA).
6

The pressure for the upper range value (20 mA
value) is present at the device, here 300 mbar (4.5psi) for example.
Select GET URV parameter.
Confirm value present. The pressure value present is
assigned to the upper current value (20 mA).

7

Result: The measuring range is configured for 0 to+300 mbar (4.5 psi).

You can also specify a customer-specific unit. See parameter description for PRESS. ENG.
UNIT (Page 16).

4.2

Calibration without reference pressure

Example:In this example, a device with a 400 mbar (6 psi) sensor is configured for the 0 to +300 mbar
(4.5 psi) measuring range, i.e. 0 mbar is assigned to the 4 mA value and 300 mbar (4.5 psi)to the 20 mA value.
Prerequisite:• This is a theoretical calibration, i.e. the pressure values for the
lower range and upper range value are known.
• See also Operating Instructions for Cerabar S (BA00412P), "Pressure measurement"
section.

Endress+Hauser

9Pressure measurement

Cerabar S 4 to 20mA HART

• Due to the orientation of the device, there may be a shift in the measured value, i.e. when
the container is empty or partially filled, the MEASURED VALUE parameter does not
display zero.  For information on how to perform position adjustment, see also Page 14,
Table 3: Position adjustment.
• For a description of the parameters mentioned, see– Page 11, Table 1: MEASURING MODE
– Page 14, Table 3: POSITION ADJUSTMENT– Page 15, Table 4: BASIC SETUP.
• For a description of further relevant parameters, see– Page 18, Table 5: EXTENDED SETUP– Page 30, Table 12: PROCESS VALUES.
Description
1

If necessary, select the "Pressure" measuring mode
via the MEASURING MODE parameter.
!

WARNING

Changing the measuring mode affects the span
(URV)!
This situation can result in product overflow.
If the measuring mode is changed, the spansetting (URV) must be verified in the"CALIBRATION"  "BASIC SETUP" operating
menu and, if necessary, reconfigured!

‣

P01-PMP71xxx-19-xx-xx-xx-000

Onsite display:
Menu path: GROUP SELECTION MEASURING
MODE

I
[mA]

Digital communication
Menu path: OPERATING MENU SETTINGS 
BASIC SETUP MEASURING MODE
2

Onsite display:
Select BASIC SETUP function group. Menu path:
GROUP SELECTION OPERATING MENU 
SETTINGS BASIC SETUP

3

Select a pressure unit via the PRESS. ENG. UNIT
parameter, here mbar for example.

4

Select SET LRV parameter.

5

➁ 20

➀ 4

Enter value, here 0 mbar, for the SET LRVparameter and confirm. This pressure value isassigned to the lower current value (4 mA).

Fig. 2:

Select SET URV parameter.

1
2

0

4

p
[bar]

P01-MIDxxxxx-05-xx-xx-xx-004

Calibration without reference pressure
See Table, Step 4.
See Table, Step 5.

Enter the value for the "SET URV" parameter (here
300 mbar (4.5 psi)) and confirm. This pressure value
is assigned to the upper current value (20 mA).
6

Result:
The measuring range is configured for 0 to +300
mbar (4.5 psi).

• You can also perform calibration without reference pressure by means of the QUICK SETUP
menu. See Page 12 ff, Table 2: QUICK SETUP menu.
• You can also specify a customer-specific unit. See parameter description for PRESS. ENG.
UNIT (Page 16).

10

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

5

Description of parameters• The following tables list all the parameters as per the menu structure. Each tablecorresponds to a function group in the menu tree. The overall menu structure is illustrated
in Section 7.1.
• The menu structure for local operation and digital communication are slightly different.
The differences mainly affect the MEASURING MODE and LANGUAGE parameters.
• In the operating program or HART handheld terminal, additional parameters are
displayed. These parameters are marked accordingly .
• The menu path is indicated in the header of each table. You can use this path to get to theparameters in question.
• The menu has a different structure depending on the measuring mode selected. Thismeans that some function groups are only displayed for one measuring mode. If certain
requirements have to be met for a function group, these are listed in the first row of thetable.
• Some parameters are only displayed if other parameters are appropriately configured.
• Parameter names are written in upper case in the text.
• In the "Parameter name" column, the unique identification number (ID) of the parameteris indicated in brackets. This ID only appears on the onsite display.

Measured value

1)
GROUP SELECTION

1)LANGUAGE

1)
MEASURING MODEQUICK SETUPOPERATING MENU

2)
MEASURING MODE

1) Display via on-site display only
2) Display via digital communication

Pressure

Level (not provided
for PMP71 with MID)
P01-MIDxxxxx-19-xx-xx-EN-001

Fig. 3:

1st selection level in menu, LANGUAGE ( see Page 11, Table 1)

Table 1: GROUP SELECTION LANGUAGE – Onsite display
Parameter name

Description

LANGUAGE (079)
Options

Select the menu language for the onsite display.
• In the operating program and in the HART handheld terminal, the LANGUAGE
parameter is arranged in the DISPLAY function group.
• Select the menu language for the operating program via the "Options" menu "Settings" "Language" tab "Tool Language" field.
Options:• Deutsch• English• Français• Italiano• Español• Nederlands• Chinese (CHS)• Japanese (JPN)
Factory setting:EnglishEndress+Hauser

11Description of parameters

Cerabar S 4 to 20mA HART

Measured value

1)
GROUP SELECTION

1)LANGUAGE

1)
MEASURING MODEQUICK SETUPOPERATING MENU

2)
MEASURING MODE

Pressure
Level (not provided
for PMP71 with MID)
POS. INPUT VALUE

SET LRVSET URV

1) Display via on-site display only
2) Display via digital communication

DAMPING VALUE

P01-MIDxxxxx-19-xx-xx-EN-003

Fig. 4:

Quick Setup menu for the "Pressure" measuring mode

Table 2: (GROUP SELECTION ) QUICK SETUP "Pressure"
Parameter name

Description

This menu displays the most important parameters for the "Pressure" measuring mode.
Prerequisite:• MEASURING MODE = Pressure
Note:
See also– Page 15 ff, Table 6: BASIC SETUP– Page 18, Table 13: EXTENDED SETUP– Page 30 ff, Table 23: PROCESS VALUES– Page 8 ff, Section 4 "Pressure measurement".
MEASURING MODE
Options

Select the measuring mode.
The operating menu is structured according to the selected measuring mode.
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
If the measuring mode is changed, the span setting (URV) must be verified inthe "CALIBRATION"  "BASIC SETUP" operating menu and, if necessary,reconfigured!

‣

Prerequisite:• Digital communication
Options:• Pressure
Factory setting:• Pressure

12

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 2: (GROUP SELECTION ) QUICK SETUP "Pressure"
Parameter name

Description

POS. INPUT VALUE (563)
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure need not be known. To correct the pressure difference, youneed a reference measured value (e. g. from a reference device).
Due to the orientation of the device, there may be a shift in the measured value, i.e.
for example, when the container is empty or partially filled, the MEASUREDVALUE parameter does not display zero or the desired value.
Example:– MEASURED VALUE = 0.5 mbar (0.0075 psi)
– For the POS. INPUT VALUE parameter, specify the desired set point for the
MEASURED VALUE, e.g. 2 mbar (0.03 psi).
(MEASURED VALUEnew = POS. INPUT VALUE)– MEASURED VALUE (after entry for POS. INPUT VALUE) = 2.0 mbar ( 0.03 psi)
– The CALIB. OFFSET parameter displays the resulting pressure difference (offset)
by which the MEASURED VALUE was corrected.
The following applies: CALIB. OFFSET = MEASURED VALUEold – POS. INPUTVALUE,
here: CALIB. OFFSET = 0.5 mbar (0.0075 psi) – 2.0 mbar (0.03 psi) = – 1.5
mbar (-0.0225 psi)
– The current value is also corrected.
Factory setting:
0.0

SET LRV (245)
Entry

Set lower range value – without reference pressure.
Enter pressure value for the lower current value (4 mA).
Factory setting:
0.0 or as per order specifications

SET URV (246)
Entry

Set upper range value – without reference pressure.
Enter pressure value for the upper current value (20 mA).
Factory setting:High sensor limit (see PRESS. SENS HILIM, Page 29) or as per orderspecifications

DAMPING VALUE (247)
EntryEnter damping time (time constant ).
The damping affects the speed at which all subsequent elements, such as the onsite
display, measured value and current output react to a change in the pressure.
Input range:
0.0 to 999.0 s
Factory setting:
2.0 s or as per order specifications

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSPOSITION ADJUSTMENTDISPLAY

BASIC SETUP

EXTENDED SETUP

POS. INPUT VALUE

1) Display via
on-site display only

CALIB. OFFSET

P01-MIDxxxxx-19-xx-xx-EN-026

Fig. 5:Endress+Hauser

Function group POSITION ADJUSTMENT

13

Description of parameters

Cerabar S 4 to 20mA HART

Table 3: (GROUP SELECTION ) OPERATING MENU SETTINGS POSITION ADJUSTMENT
Parameter name

Description

Due to the orientation of the device, there may be a shift in the measured value, i.e. when the container is empty
or partially filled, the measured value does not display zero. Cerabar S provides two different options for carryingout position adjustment.
Recommendation:• The pressure difference between zero (set point) and the measured pressure need not be known.
– POS. INPUT VALUE• The pressure difference between zero (set point) and the measured pressure is known.
– CALIB. OFFSET
POS. INPUT VALUE (563)
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure need not be known. To correct the pressure difference, youneed a reference measured value (e. g. from a reference device).
Example:– MEASURED VALUE = 0.5 mbar (0.0075 psi)
– For the POS. INPUT VALUE parameter, specify the desired set point for the
MEASURED VALUE, e.g. 2.0 mbar (0.03 psi).
(MEASURED VALUEnew = POS. INPUT VALUE)– MEASURED VALUE (after entry for POS. INPUT VALUE) = 2.0 mbar ( 0.03 psi)
– The CALIB. OFFSET parameter displays the resulting pressure difference (offset)
by which the MEASURED VALUE was corrected.
The following applies: CALIB. OFFSET = MEASURED VALUEold – POS. INPUTVALUE,
here: CALIB. OFFSET = 0.5 mbar (0.0075 psi) – 2.0 mbar (0.03 psi) = – 1.5
mbar (-0.0225 psi)
– The current value is also corrected.
Factory setting:
0.0

CALIB. OFFSET (319)
Entry

Position adjustment – the pressure difference between zero (set point) and themeasured pressure is known.
Example:– MEASURED VALUE = 2.2 mbar (0.033 psi)
– Via the CALIB. OFFSET parameter, enter the value by which the MEASUREDVALUE should be corrected. To correct the MEASURED VALUE to 0.0 mbar, youmust enter the value 2.2 here.
(MEASURED VALUE new = MEASURED VALUEold – CALIB. OFFSET)– MEASURED VALUE (after entry for calib. offset) = 0.0 mbar
– The current value is also corrected.
Factory setting:
0.0

14Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSPOSITION ADJUSTMENTDISPLAY

BASIC SETUP

EXTENDED SETUP

2)
MEASURING MODE

Pressure
PRESS. ENG. UNIT*
CUSTOMER UNIT P
CUST. UNIT FACT. P

Level (not provided
for PMP71 with MID)*

SET LRVSET URV

1) Display via on-site display only
2) Display via digital communication

GET LRV

Parameters marked with an asterisk "*"* arenot always displayed. The display
depends on the settings of other parameters,
see the following Table.

GET URVDAMPING VALUE
P01-MIDxxxxx-19-xx-xx-EN-005

Fig. 6:BASIC SETUP function group for the "Pressure" measuring mode

Table 4: (GROUP SELECTION ) OPERATING MENU SETTINGS BASIC SETUP "Pressure"
Parameter name

Description

Prerequisite:• MEASURING MODE = Pressure
Note:
See also– Page 12, Table 3: QUICK SETUP– Page 18, Table 13: EXTENDED SETUP– Page 30 ff, Table 23: PROCESS VALUES– Page 8 ff, Section 4 "Pressure measurement".
MEASURING MODE
Options

Select the measuring mode.
The operating menu is structured according to the selected measuring mode.
!

WARNING

Changing the measuring mode affects the span (URV)!
This situation can result in product overflow.
If the measuring mode is changed, the span setting (URV) must be verified inthe "CALIBRATION"  "BASIC SETUP" operating menu and, if necessary,reconfigured!

‣

Prerequisite:• Digital communication
Options:• Pressure
Factory setting• Pressure

Endress+Hauser

15Description of parameters

Cerabar S 4 to 20mA HART

Table 4: (GROUP SELECTION ) OPERATING MENU SETTINGS BASIC SETUP "Pressure"
Parameter name

Description

PRESS. ENG. UNIT (060)
Options

Select pressure unit.
If a new pressure unit is selected, all pressure-specific parameters are converted
and displayed with the new unit.
Options:• mbar, bar• mmH2O, mH2O, inH2O, ftH2O• Pa, hPa, kPa, MPa• psi• mmHg, inHg
• Torr• g/cm2, kg/cm2• lb/ft2• atm• gf/cm2, kgf/cm2• User unit See also the following parameter description for CUSTOMER UNIT P
and CUST.UNIT FACT.P.
Factory setting:
Depends on the sensor nominal measuring range mbar or bar or as per orderspecifications

CUSTOMER UNIT P (075)
EntryEnter text (unit) for customer-specific pressure unit.
You can enter a maximum of eight alphanumeric characters here. See also
CUST.UNIT FACT.P.
Prerequisite:• PRESS. ENG. UNIT = User unit
Only the first five characters are shown on the onsite display. For example, if"crates" is specified as the customer-specific unit, "crate" is displayed.
If the unit contains a slash, up to eight characters can be shown on the onsite
display. The maximum number of characters in the counter is again limited to five.
For example, if "crates/m2" is specified as the customer-specific unit, "crate/m2" isdisplayed. In FieldCare, all eight characters are displayed.
In the HART handheld terminal, the customer-specific unit is only displayed in the
CUSTOMER UNIT P parameter. The measured value is displayed with theadditional text "User Unit".
Factory setting:________

CUST.UNIT FACT.P (317)
EntryEnter conversion factor for a customer-specific pressure unit.
The conversion factor must be entered in relation to the SI unit "Pa". See also
CUSTOMER UNIT P.
Prerequisite:• PRESS. ENG. UNIT = User unit
Example:– You want the measured value to be displayed in "PU" (PU: packing unit).
– MEASURED VALUE =10,000 Pa i 1 PU
– Entry CUSTOMER UNIT P: PU– Entry CUST.UNIT FACT.P: 0.0001– Result: MEASURED VALUE = 1 PU
Factory setting:
1.0

SET LRV (245)
Entry

Set lower range value – without reference pressure.
Enter pressure value for the lower current value (4 mA).
Factory setting:
0.0 or as per order specifications

SET URV (246)
Entry

Set upper range value – without reference pressure.
Enter pressure value for the upper current value (20 mA).
Factory setting:High sensor limit ( see PRESS. SENS HILIM, Page 29)

16

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 4: (GROUP SELECTION ) OPERATING MENU SETTINGS BASIC SETUP "Pressure"
Parameter name

Description

GET LRV (309)
Entry

Set lower range value – reference pressure is present at device.
The pressure for the lower current value (4 mA) is present at device. With the"Confirm" option, you assign the lower current value to the pressure value present.
Onsite display: the pressure value present is displayed in the bottom line.
Options:• Abort• Confirm

GET URV (310)
Entry

Set upper range value – reference pressure is present at device.
The pressure for the upper current value (20 mA) is present at device. With the"Confirm" option, you assign the upper current value to the pressure value present.
Onsite display: the pressure value present is displayed in the bottom line.
Options:• Abort• Confirm

DAMPING VALUE (247)
EntryEnter damping time (time constant ).
The damping affects the speed at which all subsequent elements, such as the onsite
display, measured value and current output react to a change in the pressure.
Input range:
0.0 to 999.0 s
Factory setting:
2.0 s or as per order specifications

Endress+Hauser

17Description of parameters

Cerabar S 4 to 20mA HART

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSPOSITION ADJUSTMENTDISPLAY

BASIC SETUP

EXTENDED SETUP

Pressure
TEMP. ENG. UNIT

Level (not provided
for PMP71 with MID)

1) Display via on-site display only
Parameters marked with an asterisk "*"* arenot always displayed. The display
depends on the settings of other parameters,
see the following Table.
P01-MIDxxxxx-19-xx-xx-EN-012

Fig. 7:

EXTENDED SETUP function groupfor measuring mode "Pressure", see Page 18, Table 5

Table 5: (GROUP SELECTION ) OPERATING MENU SETTINGS EXTENDED SETUP "Pressure"
Parameter name

Description

Prerequisite:• MEASURING MODE = Pressure
Note:
See also Page 8 ff, Section 4 "Pressure measurement".

‣

TEMP. ENG. UNIT (318)
Options

Select the unit for the temperature measured values.
See also PCB TEMPERATURE (Page 26) and SENSOR TEMP. (Page 31).
Options:
• °C
• °F
• K
• R
Factory setting:°C

18

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSPOSITION ADJUSTMENTDISPLAYTOTALIZER SETUPMENU DESCRIPTION

MAIN DATA FORMATALTERNATE DATA

2)1) Display via on-site display only
2) Display via digital communication

LANGUAGEDISPLAY CONTRAST

DIGITS SETS
P01-xxxxxxxx-19-xx-xx-xx-093

Fig. 8:DISPLAY group

Table 6: (GROUP SELECTION ) OPERATING MENU DISPLAY
Parameter name

Description

MENU DESCRIPTOR (419)
Options

Specify contents for the main line of the onsite display in the measuring mode.
See also Operating Instructions BA00412P, Section "Onsite display".
Options:• Main measured value (PV)• Main measured value (%)• Pressure• Level• Tank content• Current• Temperature• Error number
The selection depends on the measuring mode chosen.
Factory setting:Main measured value (PV)

MAIN DATA FORMAT(688)
Options

Specifies the number of places after the decimal point for the value displayed in themain line.
See also Operating Instructions BA00412P, Section "Onsite display".
Options:• Auto• x.x• x.xx• x.xxx• x.xxxx• x.xxxxx
Factory setting:
Auto

ALTERNATE DATA (423)
Options

Switch on "Alternating display" mode.
In this display mode, the onsite display alternates between the following measured
values depending on the measuring mode selected.
– Pressure: primary value (PV), pressure, temperature and current
Options:• Off• On
Factory setting:
Off

Endress+Hauser

19Description of parameters

Cerabar S 4 to 20mA HART

Table 6: (GROUP SELECTION ) OPERATING MENU DISPLAY
Parameter name

Description

LANGUAGE
Selection

Select the menu language for the onsite display.
• In the case of local operation, the LANGUAGE parameter is arranged directly
under the GROUP SELECTION (menu path: GROUP SELECTION LANGUAGE,
see also Page 11).
• Select the menu language for the operating program via the "Options" menu "Settings" "Language" tab "Tool Language" field.
Options:• Deutsch• English• Français• Italiano• Español• Nederlands• Chinese (CHS)• Japanese (JPN)
Factory setting:EnglishDISPLAY CONTRAST (339) Adjust contrast of onsite display.
Entry
You specify the contrast of the display with a number. Changes are only accepted as
single steps, i.e. to change the value from "8" to "4", you need to save four times. You
can also adjust the contrast of the display by means of the keys on the electronic
insert or at the device.
See also Operating Instructions BA00412P, the section on operating keys
function.
Input range:
4...13, 4: contrast weaker (brighter), 13: contrast stronger (darker).
Factory setting:
8
DIGITS SET (840)
Display

20

This parameter is used to verify correct display of characters and digits on the userinterface. If the characters and digits are correctly displayed, this parameter shows
the string "0123456789.-".

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSPOSITION ADJUSTMENTOUTPUTTOTALIZER SETUPOUTPUT CURRENT

CURR. CHARACT.

OUTPUT FAIL MODE

ALT. CURR. OUTPUT

SET MAX. ALARM

SET MIN. CURRENTASSIGN CURRENT

Level measuringmode only

1) Display via on-site display only
P01-MIDxxxxx-19-xx-xx-EN-013

Fig. 9:Group OUTPUT

Table 7: (GROUP SELECTION ) OPERATING MENU OUTPUT

Endress+Hauser

Parameter name

Description

OUTPUT CURRENT (254)
Display

Displays the current current value.

21

Description of parameters

Cerabar S 4 to 20mA HART

Table 7: (GROUP SELECTION ) OPERATING MENU OUTPUT
Parameter name

Description

CURR. CHARACT.
(694), (695), (696), (764)
Options

Select curve of current output.
Options:

I 20 mA

➃

➂

➀

➁

➃

➁

4 mA
p
0

LRV

URV
P01-xxxxxxxx-05-xx-xx-xx-009

Fig. 10:
1
2
3
4
LRV
URV
I
p

Illustration of current output curves
Linear: lower range value = 4 mA, upper range value = 20 mA
Bi-linear: lower range value = 4 mA, center or zero = 20 mA,
upper range value = 4 mA
Linear inverse: lower range value = 20 mA, upper range value = 4 mA
Bi-linear inverse: lower range value = 20 mA, center or zero = 4 mA,
upper range value = 20 mA
Lower Range Value
Upper Range Value
Current
Measured value (pressure)The 3-digit ID number on the onsite display depends on the MEASURING MODE
selected:– (694): MEASURING MODE "Pressure"
Factory setting:LinearOUTPUT FAIL MODE (388) Select the current value in the event of an alarm.
Entry
In the event of an alarm, the current assumes the current value specified with thisparameter.
Options:• Max. alarm (110%): can be set between 21...23 mA• Hold meas. value: last measured value is kept.
• Min. alarm (–10%): 3.6 mA
See also this table SET MAX. ALARM and Operating Instructions BA00412P,
Section "Configuring current output for an alarm".
Factory setting:Max. alarm 110% (22 mA)
ALT.CURR.OUTPUT (597)
Options

Set current output if sensor limits undershot or overshot.
Options:• Normal: the current output assumes the value set via the OUTPUT FAIL MODE
and SET MAX. ALARM parameters.
• NAMUR:– Lower sensor limit undershot (E120):
Current output = 3.6 mA– Upper sensor limit overshot (E115): current output assumes the value set via
the SET MAX. ALARM parameter
Factory setting:Normal22

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 7: (GROUP SELECTION ) OPERATING MENU OUTPUT
Parameter name

Description

SET MAX. ALARM (342)
EntryEnter current value for maximum alarm current.
See also OUTPUT FAIL MODE.
Input range:
21...23 mA
Factory setting:
22 mA

SET MIN. CURRENT (343)
EntryEnter lower current limit.
Some switching units do not accept current values lower than 4.0 mA.
Options:• 3.8 mA
• 4.0 mA
Factory setting:
3.8 mA

Endress+Hauser

23Description of parameters

Cerabar S 4 to 20mA HART

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENU

TRANSMITTER INFOSETTINGSPOSITION ADJUSTMENTTOTALIZER SETUP

HART DATATRANSMITTER DATAPROCESS CONNECTIONSENSOR DATENBUS ADRESSDEVICE SERIAL No

Pmax PROC. CONN.

SENSOR SER. No.

DEVICE TYPE

ELECTR. SERIAL No

PROC. CONN TYPE

PRESS. SENS LOLIM

CUST. TAG NUMBER

MAT. PROC. CONN. +PRESS: SENS HILIMLONG TAG NUMBER

MAT. PROC. CONN. –MINIMUM SPAN

ADDITIONAL INFO.

SEAL TYPESENSOR MEAS. TYPE

DEVICE REVISION2)BURST MODE

2)
BURST OPTION

2)PREAMBLE NUMBER

DEVICE DESIGN.

MAT. MEMBRANE

BOLTS MATERIAL

2)
MANUFACTOR ID

HARDWARE REV.

FILLING FLUIDNUTS MATERIAL

2)
HART MESSAGE

SOFTWARE VERSION

Tmin SENSOR

DRAIN VENT MAT.

2)
HART DATECONFIG RECORDERPRIMARY VALUE IS

2)
PCB TEMPERATURE

2)ALLOWED MIN. TEMPMOUNTING THREAD

ALLOWED MAX. TEMPREMOTE SEAL +DIP STATUS

REMOTE SEAL –2)

2)
SECONDARY VAL. IS

2)2)
SECONDARY VALUE2)2)
THIRD VALUE IS

DIAPHRAGM SEAL +2)2)

DIAPHRAGM SEAL –THIRD VALUE

2)2)

NR OF REMOTE SEAL

4TH VALUE IS

1) Display via on-site display only
2) Display via digital communication

SENS H/WARE REV.

THREAD2)
PRIMARY VALUE

Tmax SENSOR

DRAIN VENT POS.

2)

2)
4 TH VALUE

2)
FILL FLUID
P01-xxxxxxxx-19-xx-xx-xx-095

Fig. 11:

TRANSMITTER INFO group
For the HART DATA function group, see Page 24, Table 8
For the TRANSMITTER DATA function group, see Page 26, Table 9
For the PROCESS CONNECTION function group, see Page 27, Table 10
For the SENSOR DATA function group, see Page 29, Table 11

Table 8: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO HART DATA
Parameter name

Description

BUS ADDRESS (345)
EntryEnter the address for the exchange of data with the HART protocol.
(HART 5.0: range 0...15, HART 6.0: range 0...63)
Factory setting:0

DEVICE TYPE (802)Display

Displays the device identification number in decimal numerical format,
here Cerabar S: 24
Prerequisite:• Cerabar S pressure transmitter

DEVICE REVISION (699)
Display24

Displays the device revision

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 8: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO HART DATA
Parameter name

Description

BURST MODE
Options

Switch the burst mode on and off.
Options:• On• Off
Prerequisite:• Digital communication

BURST OPTION
Entry

Use this parameter to specify what command is sent to the master.
Prerequisite:• Digital communication
Factory setting:
3 (HART command 3)

PREAMBLE NUMBER
(036)
Entry

Enter the number of preambles in the HART protocol. (Synchronization of themodem modules along a transmission path, each modem module could "swallow" abyte - at least 2 bytes must arrive.)
Input range:
2...20
Factory setting:
5

MANUFACTOR ID (432)Display

Displays the manufacturer number in a decimal numerical format.
Here: 17 Endress+Hauser

HART MESSAGE (271)
EntryEnter a message (max. 32 alphanumeric characters).
On command from the master, this message is sent via the HART protocol.
Factory setting:________________________________or as per order specifications

HART DATE (481)
EntryEnter the date of the last configuration change.

PRIMARY VALUE IS
DisplayThis parameter displays the following measured value depending on the measuringmode selected:– Measuring mode "Pressure": PRESSURE

Factory setting:
DD.MM.YY (date of final test)

See also PRIMARY VALUE.
Prerequisite:• Digital communication
PRIMARY VALUE
Display

Display of primary value.
See also PRIMARY VALUE IS.
Prerequisite:• Digital communication

SECONDARY VAL. IS

Select second process value.
You can choose between the following process values depending on the measuringmode selected:– PRESSURE– CORRECTED PRESS.
– SENSOR PRESSURE– SENSOR TEMP.
– PCB TEMPERATURE– LEVEL BEFORE LIN– TANK CONTENT
Prerequisite:• Digital communication

SECONDARY VALUE

Display second process value. See also SECONDARY VAL. IS.
Prerequisite:• Digital communication

Endress+Hauser

25Description of parameters

Cerabar S 4 to 20mA HART

Table 8: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO HART DATA
Parameter name

Description

THIRD VALUE IS

Select third process value. See also . SECONDARY VAL. IS.
Prerequisite:• Digital communication

THIRD VALUE

Display third process value. See also SECONDARY VAL. IS.
Prerequisite:• Digital communication

4TH VALUE IS

Select fourth process value. See also . SECONDARY VAL. IS.
Prerequisite:• Digital communication

4TH VALUE

Display fourth process value. See also SECONDARY VAL. IS.
Prerequisite:• Digital communication

Table 9: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO TRANSMITTER DATA
Parameter name

Description

DEVICE SERIAL No (354)
Display

Displays the serial number of the device (11 alphanumeric characters).

ELECTR. SERIAL No (386)Display

Displays the serial number of the main electronics (11 alphanumeric characters).

CUST. TAG NUMBER (055) Enter device tag e.g. TAG number (max. 8 alphanumeric characters).
Entry
Factory setting:_ _ _ _ _ _ _ _ or as per order specifications
LONG TAG NUMBER (305) Enter device tag e.g. TAG number (max. 32 alphanumeric characters).
Entry
Factory setting:________________________________or as per order specifications
ADDITIONAL INFO. (272)
EntryEnter the tag description (max. 16 alphanumeric characters).

DEVICE DESIGN. (350)
Display

Displays the device designation and order code.

HARDWARE REV. (266)
Display

Displays the revision number of the main electronics
e.g.: V02.00

SOFTWARE VERSION(264)
Display

Displays the software version
V02.10.54

CONFIG RECORDER (352)
Display

Displays the configuration counter.
This counter is increased by one with each change to a parameter or group. The
counter counts to 65535 and then starts again at zero. Changes in the parametersof the DISPLAY function group do not increase the counter.

Factory setting:_ _ _ _ _ _ _ _ _ _ _ _ _ _ _ _ or as per order specifications

PCB TEMPERATURE (357) Displays the measured temperature of the main electronics.
Display

26

ALLOWED MIN. TEMP
(358)
Display

Displays the lower temperature limit of the main electronics.

ALLOWED MAX. TEMP
(359)
Display

Displays the upper temperature limit of the main electronics.

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 9: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO TRANSMITTER DATA
Parameter name

Description

DIP STATUS (363)
Display

Displays the status of DIP switch 1 on the electronic insert.
You can lock or unlock parameters relevant to the measured value with DIP switch
1. If operation is locked by means of the INSERT PIN No. parameter, you can only
unlock operation again by means of this parameter. (INSERT PIN NO, see Page
33.)
See also Operating Instructions BA00412P, "Locking/unlocking operation".
Display:• On (locking switched on)• Off (locking switched off)
Factory setting:
Off (locking switched off)

Table 10: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO PROCESS CONNECTION
Parameter name

Description

Pmax PROC. CONN. (570)
Entry

For entering and displaying the maximum permitted pressure of the processconnection.
Factory setting:as per nameplate data (see also Operating Instructions BA00412P,
Section "Nameplate")

PROC. CONN. TYPE (482)
Options

For selecting and displaying the process connection type.

MAT. PROC. CONN. +(360)
Options

For selecting and displaying the material of the process connection (P+).
See also parameter description for MAT. PROC. CONN. -Options:• Not used
• Unknown• Special• Oval flange• Thread female• Thread male
• Flange• Remote seal

Options:• Not used
• Unknown• Special• Steel• 304 st. steel• 316 st. steel• Alloy C• Monel• Tantalum• Titanium• PTFE (Teflon)• 316L st. steel• PVC• Inconel• PVDF• ECTFE
Factory setting:As per order specifications

Endress+Hauser

27Description of parameters

Cerabar S 4 to 20mA HART

Table 10: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO PROCESS CONNECTION
Parameter name

Description

SEAL TYPE (362)
Options

For selecting and displaying the material of the process seal.
Options:• Not used
• Unknown• Special• FKM Viton• NBR• EPDM• Urethane• IIR• Kalrez• FKM Viton oxyg
• CR
• MVQ
• PTFE glass• PTFE graphite• PTFE oxygen• Copper• Copper f. oxygen
Factory setting:As per order specifications

BOLTS MATERIAL

For selecting and displaying the material of the bolts.
Prerequisite:• Digital communication

NUTS MATERIAL

For selecting and displaying the material of the nuts.
Prerequisite:• Digital communication

DRAIN VENT MAT.

For selecting and displaying the material of the vent valves.
Prerequisite:• Digital communication

DRAIN VENT POS.

For selecting and displaying the position of the vent valves.
Prerequisite:• Digital communication

THREAD PROCESSFor selecting and displaying the process connection thread.
Prerequisite:• Digital communication

MOUNTING THREAD

For selecting and displaying the ways of securing the device.
Prerequisite:• Digital communication

REMOTE SEAL +For selecting and displaying the diaphragm seal type on the positive side.
Prerequisite:• Digital communication

REMOTE SEAL -For selecting and displaying the diaphragm seal type on the negative side.
Prerequisite:• Digital communication

DIAPHRAG. MAT. +For selecting and displaying the diaphragm material on the positive side
Prerequisite:• Digital communication

DIAPHRAG. MAT. -For selecting and displaying the diaphragm material on the negative side.
Prerequisite:• Digital communication

NR OF REMOTE SE

For selecting and displaying the number of diaphragm seals.
Prerequisite:• Digital communication

28

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 10: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO PROCESS CONNECTION
Parameter name

Description

FILL FLUID

For selecting and displaying the diaphragm seal fill fluid.
Prerequisite:• Digital communication

Table 11: (GROUP SELECTION ) OPERATING MENU TRANSMITTER INFO SENSOR DATA
(all measuring modes)
Parameter name

Description

SENSOR SER. No. (250)Display

Displays the serial number of the sensor (11 alphanumeric characters).

PRESS. SENS LOLIM (484)Display

Displays the lower measuring limit of the sensor.

PRESS. SENS HILIM (485)Display

Displays the upper measuring limit of the sensor.

MINIMUM SPAN (591)
Display

Displays the smallest possible span.

SENSOR MEAS.TYPE (581) Displays the sensor type.
Display• Cerabar S with absolute pressure sensor = Absolute

Endress+Hauser

MAT. MEMBRANE (365)
Display

Displays the material of the process isolating diaphragm.

FILLING FLUID (366)
Display

Displays the filling fluid.

Tmin SENSOR (368)
Display

Displays the lower nominal temperature limit of the sensor.

Tmax SENSOR (369)
Display

Displays the upper nominal temperature limit of the sensor.

SENS H/WARE REV (487)
Display

Displays the revision number of the sensor hardware.
e.g.: 1

Factory setting:as per version in order code
See Technical Information for Cerabar S TI00383P, Section "Ordering
information".

29

Description of parameters

Cerabar S 4 to 20mA HART

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSPOSITION ADJUSTMENTPROCESS INFOTOTALIZER SETUP

PROCESS VALUE

Pressure
2)
MEASURED VALUE

Level (not provided
for PMP71 with MID)

PRESSURE

CORRECTED PRESS.

SENSOR PRESSURESENSOR TEMP.

MEAS. VAL. TREND

1) Display via on-site display only
2) Display via digital communication
P01-MIDxxxxx-19-xx-xx-EN-014

Fig. 12:PROCESSINFO group
For function group PROCESS VALUES, measuring mode "Pressure", see Page 30, Table 12

Table 12: (GROUP SELECTION ) OPERATING MENU PROCESSINFO PROCESS VALUES "Pressure"
Parameter name

Description

Prerequisite:• MEASURING MODE = Pressure
MEASURED VALUE (679)

Displays the measured value
In the "Pressure" measuring mode, this value corresponds to the PRESSURE
parameter.
Prerequisite:• Digital communication
Onsite operation:• In the case of local operation, the MEASURED VALUE parameter is displayed on
the top level.

30

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 12: (GROUP SELECTION ) OPERATING MENU PROCESSINFO PROCESS VALUES "Pressure"
Parameter name

Description

PRESSURE (301)
Display

Displays the measured pressure after sensor recalibration, position adjustment anddamping. This value corresponds to the MEASURED VALUE parameter in the"Pressure" measuring mode.

SensorSensortrim

SENSOR
PRESSURE

Positionadjustment

Damping

P

PressureI

Current
outputCORRECTED PRESSURE
PRESS.
P01-MIDxxxxx-05-xx-xx-EN-001

Endress+Hauser

CORRECTED PRESS. (434)Display

Displays the measured pressure after sensor trim and position adjustment andbefore damping. See also PRESSURE diagram.

SENSOR PRESSURE (584)
Display

Displays the measured pressure before sensor trim, position adjustment anddamping.  See also PRESSURE diagram.

SENSOR TEMP. (367)
Display

Displays the temperature currently measured in the sensor. This can deviate fromthe process temperature.

MEAS. VAL. TREND (378)
Display

Displays the trend of the pressure measured value.
Possibilities: increasing, decreasing, constant31Description of parameters

Cerabar S 4 to 20mA HART

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSOPERATIONDIAGNOSTICSPOSITION ADJUSTMENTSIMULATIONMESSAGES

USER LIMITSENTER RESET CODESIMULATION MODE

ALARM STATUS

Pmin ALARM WINDOWOPERATING HOURSILAST DIAG. CODE

Pmax ALARM WINDOWINSERT PIN NO

J

ACK. ALARM MODE

HistoROM AVAIL.

1) Display via on-site display only
2) Display via digital communication

ACK. ALARM*RESET ALL ALARMS*ERROR No.

DOWNLOAD SELECT.

HistoROM CONTROL

Tmin ALARM WINDOW*

Tmax ALARM WINDOW

2)Parameters marked with an asterisk "*"* arenot always displayed. The display

SELECT ALARMTYPE

depends on the settings of other parameters,
see the following Table.

ALARM DELAY

ALARM DISPL. TIME

I
none

Pressure

Tank content

Alarm Warning

Current

SIM. PRESSURE

SIM. TANK CONT.

SIM. CURRENT

SIM. ERROR NO.

J

J

J

J

J

P01-MIDxxxxx-19-xx-xx-EN-015

Fig. 13:OPERATING and DIAGNOSTICS group
For the OPERATING group, see Page 32, Table 13
For the SIMULATION MODE function group, see Page 34, Table 14
For the MESSAGES function group, see Page 35, Table 15
 For the USER LIMITS function group, see Page 36, Table 16

Table 13: (GROUP SELECTION ) OPERATING MENU OPERATING
Parameter name

Description

ENTER RESET CODE (047)
Entry

Reset parameters completely or partially to factory values or delivery status.
See also Operating Instructions BA00412P, Section "Factory setting (reset)".
Factory setting:
0

OPERATING HOURS (409)
Display32

Displays the hours of operation. This parameter cannot be reset.

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 13: (GROUP SELECTION ) OPERATING MENU OPERATING
Parameter name

Description

INSERT PIN NO (048)
Entry

For entering a code to lock or unlock operation.
• The
symbol on the onsite display indicates that operation is locked.
Parameters which refer to how the display appears, e.g. LANGUAGE andDISPLAY CONTRAST can still be altered.
• If operation is locked by means of the DIP switch, you can only unlock operation
again by means of the DIP switch. If operation is locked by means of the onsite
display or remote operation e.g. FieldCare, you can unlock operation again either
by means of the onsite display or remote operation.
See also Operating Instructions BA00412P, "Locking/unlocking operation".
Options:• Lock: enter a number between 0 and 9999 that is 100.
• Unlock: enter the number 100.
Factory setting:100

HistoROM AVAIL. (831)
Display

Indicates whether the optional HistoROM®/M-DAT memory module is connectedto the electronic insert.
See also Operating Instructions BA00412P, Section "HistoROM®/M-DAT(optional)".
Options:• Yes (HistoROM®/M-DAT is attached to the electronic insert)• No (HistoROM®/M-DAT is not attached to the electronic insert)

DOWNLOAD SELECT(014)
Options

Select download function from HistoROM to device. The option selected has noeffect on an upload from the device to the HistoROM.
Prerequisite:• A HistoROM®/M-DAT is attached to the electronic insert(HistoROM AVAIL. = Yes)
Options:• Configuration copy: With this option, all parameters apart from the
TRANSMITTER SERIAL No, DEVICE DESIGN., CUST. TAG NUMBER, LONG TAG
NUMBER, ADDITIONAL INFO., BUS ADDRESS and the parameters of the
POSITION ADJUSTMENT and PROCESS CONNECTION group are overwritten.
• Device replacement: With this option, all parameters except for TRANSMITTERSERIAL No, DEVICE DESIGNATION and the parameters of the POSITION
ADJUSTMENT and PROCESS CONNECTION group are overwritten.
• Electronics replace: With this option, all parameters except for the parameters of
the POSITION ADJUSTMENT group are overwritten.
Factory setting:Configuration copy (if HistoROM®/M-DAT is attached to the electronic insert)

HistoROM CONTROL (832) For selecting the direction for copying the data.
Options
See also Operating Instructions BA00412P, Section "HistoROM®/M-DAT(optional)".
Prerequisite:• A HistoROM®/M-DAT is attached to the electronic insert(HistoROM AVAIL. = Yes)
Options:• Abort• HistoROM Device• Device HistoROM
Factory setting:Abort (if HistoROM®/M-DAT is attached to the electronic insert)

Endress+Hauser

33

Description of parameters

Cerabar S 4 to 20mA HART

Table 14: (GROUP SELECTION ) OPERATING MENU DIAGNOSTICS SIMULATION MODE
Parameter name

Description

SIMULATION MODE (413) Switch on simulation and select simulation type.
Options
Any simulation running is switched off if the measuring mode or level type ischanged.
Options:• None• Pressure, see also this table, parameter description for SIM. PRESSURE.
• Tank content, see also this table, parameter description for SIM. TANK CONT.
• Current, see also this table, parameter description for SIM. CURRENT• Alarm/warning, , see also this table, parameter description for SIM. ERROR
NO.

SensorPositionadjustment

Sensortrim

Damping

Simulation value pressure

P

PressureI

Current
outputSimulation value
current

P01-MIDxxxxx-05-xx-xx-EN-003

Factory setting:
NoneSIM. PRESSURE (414)
EntryEnter simulation value.
See also SIMULATION MODE.
Prerequisite:• SIMULATION MODE = Pressure
Factory setting:Current pressure measured valueSIM. CURRENT (270)
Entry

Enter simulation value.
See also SIMULATION MODE.
Prerequisite:• SIMULATION MODE = Current value
Factory setting:Current current valueSIM. ERROR NO. (476)
EntryEnter message number.
See also SIMULATION MODE.
See also these Operating Instructions, Section 6.1 "Messages", "Code" table
column.
Prerequisite:• SIMULATION MODE = Alarm/warning
Factory setting:
613 (simulation active)

34

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 15: (GROUP SELECTION ) OPERATING MENU DIAGNOSTICS MESSAGESParameter name

Description

ALARM STATUS (046)
Display

Displays the current messages present. See also these Operating Instructions,
Section 6.1 "Messages" and Section "Confirming messages".
Onsite display• The measured value display shows the message with the highest priority.
• The ALARM STATUS parameter shows all the messages in descending order ofpriority. You can scroll through all the messages present with the O or S key.
Operating program• The "Status" field and the ALARM STATUS parameter show the message with thehighest priority.

LAST DIAG. CODE (564)
Display

Displays the last messages that occurred and were eliminated.
• Onsite display: you can scroll through the last 15 messages with the O or S key.
• Digital communication: the last message is displayed.
• Use the RESET ALL ALARMS parameter to delete the messages listed in the
LAST DIAG. CODE parameter.

ACK. ALARM MODE (401) Switch on acknowledge alarm mode.
Options
See also ACK. ALARM.
Options:• On• Off
Factory setting:
Off
ACK. ALARM (500)
Options

Acknowledge alarm.
Prerequisite:• ACK. ALARM MODE = On
Options:• Abort• Confirm
The cause of the alarm must be eliminated, the message must be acknowledged via
the ACK. ALARM parameter and, where applicable, the ALARM DISPL. TIME (Page 36) has to have elapsed before the device starts measuring again following an
alarm.
See also these Operating Instructions, Section 6.3 "Confirming messages".
Factory setting:
Abort

RESET ALL ALARMS (603) Use this parameter to reset all the messages of the LAST DIAG. CODE parameter.
Options
Options:• Abort• Confirm
Factory setting:
Abort
ERROR No.
Entry

For "Error"-type messages, you can decide whether the device should behave as in
the event of an alarm (A) or as in the event of a warning (W). Enter thecorresponding message number for this parameter. See also SELECTALARMTYPE.
See also these Operating Instructions, Section 6.1 "Messages" and Section 6.2"Response of outputs to errors".
Prerequisite:• Digital communication

Endress+Hauser

35Description of parameters

Cerabar S 4 to 20mA HART

Table 15: (GROUP SELECTION ) OPERATING MENU DIAGNOSTICS MESSAGESParameter name

Description

SELECT ALARMTYPE(595) – Entry(600) – Selection

For "Error"-type messages, you can decide whether the device should behave as in
the event of an alarm (A) or as in the event of a warning (W). See also ERROR
No.
See also these Operating Instructions, Section 6.2 "Response of outputs toerrors".
Options:• Alarm (A): output current assumes a defined value.
• Warning (W): device continues measuring
Onsite operation:
1. Enter the corresponding message number for ERROR No. field.
2. Select "Alarm" or "Warning" option.
Digital communication
1. Enter the corresponding message number via the ERROR No. parameter.
2. Use the SELECT ALARMTYPE parameter to select the "Alarm" or "Warning"
option.

ALARM DELAY (336)
EntryEnter the alarm response time for all "Error"-type messages.
There is no alarm if the cause of the error is eliminated within the alarm delay
time.
Input range:
0...100 s
Factory setting:
0.0 s

ALARM DISPL. TIME (480) Enter the alarm delay time for all "Error"-type messages. Once the cause of the error
Entry
is rectified, the alarm display time starts running.
The following applies if the setting for ACK. ALARM MODE = On
If an alarm appears and the alarm display time elapses before the alarm has been
acknowledged, the message will be cleared once it has been acknowledged.
See also these Operating Instructions, Section 6.3 "Confirming messages".
Input range:
0 to 999.9 s
Factory setting:
0.0 s

Table 16: (GROUP SELECTION ) OPERATING MODE DIAGNOSTICS USER LIMITSParameter name

Description

Pmin ALARM WINDOW
(332)
Entry

Customer-specific process monitoring – enter lower pressure limit.
You can use the SELECT ALARMTYPE parameter to enter how the device responds
if the operating pressure undershoots the specified value.
See also these Operating Instructions, Section 6.1 "Messages", Table, code E730
and Section 6.2 "Response of outputs to errors".
Factory setting:
Low sensor limit •1.1
(For the low sensor limit, see PRESS. SENS LOLIM.)

Pmax ALARM WINDOW
(333)
Entry

Customer-specific process monitoring – enter upper pressure limit.
You can use the SELECT ALARMTYPE parameter to enter how the device responds
if the operating pressure undershoots the specified value.
See also these Operating Instructions, Section 6.1 "Messages", Table, code E731
and Section 6.2 "Response of outputs to errors".
Factory setting:High sensor limit • 1.1(For the high sensor limit, see PRESS. SENS HILIM.)

36

Endress+Hauser

Cerabar S 4 to 20mA HART

Description of parameters

Table 16: (GROUP SELECTION ) OPERATING MODE DIAGNOSTICS USER LIMITSParameter name

Description

Tmin ALARM WINDOW
(334)
Entry

Customer-specific process monitoring – enter lower temperature limit.
You can use the SELECT ALARMTYPE parameter to enter how the device responds
if the operating pressure undershoots the specified value.
See also these Operating Instructions, Section 6.1 "Messages" Table, code E732
and Section 6.2 "Response of outputs to errors".
Factory setting:Lower sensor temperature application limit – 10 K
(For the lower temperature application limit, see Tmin SENSOR)

Tmax ALARM WINDOW
(335)
Entry

Customer-specific process monitoring – enter upper temperature limit.
You can use the SELECT ALARMTYPE parameter to enter how the device responds
if the operating pressure undershoots the specified value.
See also these Operating Instructions, Section 6.1 "Messages" Table, code E733
and Section 6.2 "Response of outputs to errors".
Factory setting:Upper sensor temperature application limit +10 K(For the upper temperature application limit, see Tmax SENSOR)

Measured value

1)
GROUP SELECTION

1)LANGUAGE

OPERATING MENUSETTINGSSERVICEPOSITION ADJUSTMENTSYSTEM 2

CURR. TRIM 4mA

CURR. TRIM 20mA

OFFSET 4mA TRIM

1) Display via on-site display only

OFFSET 20mA TRIM

P01-MIDxxxxx-19-xx-xx-EN-016

Fig. 14:

Endress+Hauser

SYSTEM 2 group37

Description of parameters

Cerabar S 4 to 20mA HART

Table 17: (GROUP SELECTION ) OPERATING MENU SERVICE SYSTEM 2Parameter name

Description

CURR. TRIM 4mA (045)
EntryEnter the current value for the lower point (4 mA) of the current linear regression
line.
You can adapt the current output to the transmission conditions with thisparameter and "CURR. TRIM 20mA".
Perform the current trim for the lower point as follows:
1. Select the SIMULATION MODE group.
(Menu path: (GROUP SELECTION) OPERATING MENU DIAGNOSTICS 
SIMULATION MODE)
2. Select the "Current" option via the SIMULATION MODE parameter.
3. Enter "4 mA" for the SIM. CURRENT parameter.
4. Select the SYSTEM 2 group.
(Menu path: (GROUP SELECTION) OPERATING MENU SERVICE)
5. Enter the current value measured with the switching unit in the "CURR. TRIM
4mA" parameter.
Input range:
Measured current ±0.2 mA
Factory setting:
4 mA

CURR. TRIM 20mA (042)
EntryEnter the current value for the upper point (20 mA) of the current linear regression
line.
You can adapt the current output to the transmission conditions with thisparameter and "CURR. TRIM 4mA".
Perform the current trim for the upper point as follows:
1. Select the SIMULATION MODE group.
(Menu path: (GROUP SELECTION) OPERATING MENU DIAGNOSTICS 
SIMULATION MODE)
2. Select the "Current" option via the SIMULATION MODE parameter.
3. Enter "20 mA" for the SIM. CURRENT parameter.
4. Select the SYSTEM 2 group.
(Menu path: (GROUP SELECTION) OPERATING MENU SERVICE)
5. Enter the current value measured with the switching unit in the "CURR. TRIM
20mA" parameter.
Input range:
Measured current ±0.2 mA
Factory setting:
20 mA

OFFSET 4mA TRIM (043)
Display

Displays the difference between 4 mA and the value entered for the "CURR. TRIM
4mA" parameter.
Factory setting:0

OFFSET 20mA TRIM (044) Displays the difference between 20 mA and the value entered for the "CURR TRIM
Display
20mA" parameter.
Factory setting:038

Endress+Hauser

Cerabar S 4 to 20mA HART

Troubleshooting

6

Troubleshooting

6.1

Messages

The following table lists all the possible messages that can occur.
The device makes a distinction between the error types "Alarm", "Warning" and "Error". You
may specify whether the device should react as if for an "Alarm" or "Warning" for "Error"
messages.
 See "Error type/NA 64" column and parameter description for ERROR No. and SELECTALARMTYPE (Page 36).
In addition, the "Error type/NA 64" column classifies the messages in accordance with
NAMUR Recommendation NA 64:• Break down: indicated with "B"• Maintenance need: indicated with "C" (check request)• Function check: indicated with "I" (in service)
Error message display on the onsite display:• The measured value display shows the message with the highest priority. See the"Priority" column.
• The ALARM STATUS parameter (Page 35) shows all the messages in descending orderof priority. You can scroll through all the messages present with the S key or O key.
Message display via digital communication:• The ALARM STATUS parameter (Page 35) shows the message with the highest
priority.
See the "Priority" column.
• If the device detects a defect in the onsite display during initialization, special error
messages are generated. For the error messages, see Page 45, Section 6.1.1 "Onsite
display error messages".
• For support and further information, please contact Endress+Hauser Service.
CodeError type/
NA 64

101 (A101) Alarm
B

106(W106)

WarningC

110 (A110) AlarmB

113 (A113) Alarm
B

Endress+Hauser

Message/description

Cause

MeasurePriority

B>Sensor electronic EEPROM
error– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P. This message
normally only appears briefly.

– Wait a few minutes.
– Restart the device. Perform reset(Code 62).
– Block off electromagnetic effects
or eliminate source ofdisturbance.

17– Sensor defective.

– Replace sensor.

– Downloading.

– Wait for download to complete.

52– Reestablish supply voltage.
Perform reset (Code 7864) if
necessary. Carry out calibration
again.

6

C>Downloading - please wait

B>Checksum error in EEPROM: – The supply voltage isconfiguration segment
disconnected when writing.

B>ROM failure in transmitterelectronic– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Block off electromagnetic effects
or eliminate sources ofdisturbance.

– Main electronics defective.

– Replace main electronics.

– Main electronics defective.

– Replace main electronics.

1

39

Troubleshooting

Cerabar S 4 to 20mA HART

Code

Error type/
NA 64

Message/description

Cause

MeasurePriority

115 (E115)

Error
B
Factory
setting:
Warning

B>Sensor overpressure– Overpressure present.

– Reduce pressure until message
disappears.

29– Sensor defective.

– Replace sensor.

116(W116)

Warning
CC>Download error, repeat
download– The file is defective.

– Use another file.

– During the download, the data
are not correctly transmitted tothe processor, e.g. because ofopen cable connections, spikes(ripple) on the supply voltage orelectromagnetic effects.

– Check cable connection PC –transmitter.
– Block off electromagnetic effects
or eliminate sources ofdisturbance.
– Perform reset (Code 7864) andcarry out calibration again.
– Repeat download.

120 (E120)

Error
B
Factory
setting:
Warning

B>Sensor low pressure– Pressure too low.

– Increase pressure until message
disappears.

– Sensor defective.

– Replace sensor.

36

30

121 (A121) Alarm
B

B>Checksum error in factory
segment of EEPROM– Main electronics defective.

– Replace main electronics.

5

122 (A122) AlarmB

B>Sensor not connected– Cable connection sensor –main
electronics disconnected.

– Check cable connection and repair 13
if necessary.

– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Main electronics defective.

– Replace main electronics.

– Sensor defective.

– Replace sensor.

130 (A130) Alarm
B

B>EEPROM is defective.

– Main electronics defective.

– Replace main electronics.

10

131 (A131) Alarm
B

B>Checksum error in EEPROM: – Main electronics defective.
min/max segment– Replace main electronics.

9132 (A132) Alarm
B

B>Checksum error in totalizer
EEPROM– Main electronics defective.

– Replace main electronics.

7

133 (A133) Alarm
B

B>Checksum error in History
EEPROM– An error occurred when writing.

– Perform reset (Code 7864) andcarry out calibration again.

8– Main electronics defective.

– Replace main electronics.

– The linearization table is not
monotonic increasing or
decreasing.

– Add to or correct linearizationtable. Then accept linearizationtable again.

602(W602)

40

Warning
C

C>Linearization curve notmonotone57

Endress+Hauser

Cerabar S 4 to 20mA HART

Troubleshooting

CodeError type/
NA 64

Message/description

Cause

Measure604(W604)

Warning
C

C>Linearization table not valid. – The linearization table consists of – Add to linearization table. If
less than 2 points.
necessary, confirm linearization
Less than 2 points or points tootable again.
close

Priority58– Correct linearization table and– At least 2 points in theaccept again.
linearization table are too close
together. A minimum gap of
0.5 % of the span must be
maintained between two points.
Spans for the "Pressure linearized"
option: HYDR. PRESS MAX. –HYDR. PRESS MIN.; TANKCONTENT MAX. – TANKCONTENT MIN.
Spans for the "Height linearized"
option: LEVEL MAX – LEVELMIN; TANK CONTENT MAX. –TANK CONTENT MIN.
613
(W613)

Warning
I

I>Simulation is active– Simulation is switched on, i.e. the – Switch off simulation.
device is not measuring atpresent.

620 (E620)

Error
C
Factory
setting:
Warning

C>Current output out of range

The current is outside the permitted
range 3.8 to 20.5 mA.

700(W700)

WarningC– The pressure applied is outside
the set measuring range (butwithin the sensor range).

– Check pressure applied,reconfigure measuring range ifnecessary.
See also these Operating
Instructions Section 4.
– Perform reset (Code 7864) andcarry out calibration again.

– Loose connection at sensor cable– Wait a short period of time andtighten the connection, or avoid
loose connection.

C>Last configuration not stored – An error occurred when writing
or reading configuration data or
the power supply was
disconnected.

– Perform reset (Code 7864) andcarry out calibration again.

– Main electronics defective.

– Replace main electronics.
– Carry out calibration again.

701(W701)

WarningC

C>Measuring chain config.
exceeds sensor range– The calibration carried out would
result in the sensor nominal
operating range being undershot
or overshot.

702(W702)

Warning
C

C>HistoROM data notconsistent.

– Data were not written correctly to – Repeat upload.
– Perform reset (Code 7864) andthe HistoROM, e.g. if thecarry out calibration again.
HistoROM was detached during
the writing process.

703 (A703) AlarmB

704 (A704) AlarmB

705 (A705) Alarm
B

Endress+Hauser

B>Measurement error

B>Measurement error

B>Measurement error

60

49

54

51

55

– HistoROM does not have anydata.

– Copy suitable data to theHistoROM. (See also Operating
Instructions BA00412P, Section"Copying configuration data".)– Fault in the main electronics.

– Briefly disconnect device from the 22
power supply.

– Main electronics defective.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 12
power supply.

– Main electronics defective.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 21
power supply.

– Main electronics defective.

– Replace main electronics.

41

Troubleshooting

Cerabar S 4 to 20mA HART

Code

Error type/
NA 64

Message/description

Cause

MeasurePriority

706(W706)

Warning
C

C>Configuration in HistoROM
and device not identical– Configuration (parameters) in the – Copy data from the device to the 59
HistoROM.
HistoROM and in the device is not– Copy data from the HistoROM toidentical.
the device. The message remains
if the HistoROM and the device
have different software versions.
The message goes out if you copy
the data from the device to theHistoROM.
– Device reset codes such as 7864
do not have any effect on the
HistoROM. That means that if you
perform a reset, theconfigurations in the HistoROM
and in the device may not be thesame.
See also Operating Instructions
BA00412P, Section "Copying
configuration data".
38

707 (A707) AlarmB

B>X-VAL. of lin. table out ofedit limits.

– Carry out calibration again. (– At least one X-VALUE in the
See also these Operating
linearization table is either below
Instructions, Section 5.)
the value for HYDR. PRESS MIN.
or MIN. LEVEL or above the valuefor HYDR. PRESS. MAX. or LEVEL
MAX

710(W710)

B>Set span too small. Not
allowed.

– Values for calibration (e.g. lower
range value and upper range
value) are too close together.

– Adjust calibration to suit sensor. 51( See also Page 29, parameter
description for MINIMUM SPAN.)– The sensor was replaced and thecustomer-specific configuration
does not suit the sensor.

– Adjust calibration to suit sensor.
– Replace sensor with a suitablesensor.

– Unsuitable download carried out.

– Check configuration and perform
download again.

WarningC

711 (A711) Alarm
B

B>LRV or URV out of edit limits – Lower range value and/or upper – Reconfigure lower range value
and/or upper range value to suitrange value undershoot or
the sensor. Pay attention to
overshoot the sensor range limits.
position factor.
– The sensor was replaced and thecustomer-specific configuration
does not suit the sensor.

– Reconfigure lower range value
and/or upper range value to suit
the sensor. Pay attention toposition factor.
– Replace sensor with a suitablesensor.

– Unsuitable download carried out.

– Check configuration and perform
download again.

713 (A713) Alarm
B

B>100% POINT level out of edit – The sensor was replaced.
limits715 (E715)

C>Sensor over temperature

716 (E716)

42

Error
C
Factory
setting:
Warning

Error
B
Factory
setting:
Alarm

B>Process isolating diaphragmbroken– Carry out calibration again.

37

39– The temperature measured in the – Reduce process temperature/
ambient temperature.
sensor is greater than the upper
nominal temperature of thesensor. (See also these
Operating Instructions, parameter
description for Tmax SENSOR.)– Unsuitable download carried out.

– Check configuration and perform
download again.

– Sensor defective.

– Replace sensor.

32

24Endress+Hauser

Cerabar S 4 to 20mA HART

Troubleshooting

CodeError type/
NA 64

Message/description

Cause

717 (E717)

Error
C
Factory
setting:
Warning

C>Transmitter overtemperature– The temperature measured in the – Reduce ambient temperature.
electronics is greater than theupper nominal temperature of
the electronics (+88 °C).

Error
C
Factory
setting:
Warning

C>Transmitter undertemperature

Error
C
Factory
setting:
Warning

C>Sensor under temperature

718 (E718)

720 (E720)– Unsuitable download carried out.

Measure34– Check configuration and perform
download again.

– The temperature measured in the – Increase ambient temperature.
Insulate device if necessary.
electronics is smaller than thelower nominal temperature of theelectronics (–43 °C).
– Unsuitable download carried out.

Priority

35– Check configuration and perform
download again.

– The temperature measured in the – Increase process temperature/
ambient temperature.
sensor is smaller than the lower
nominal temperature of thesensor. ( See also Page 29,
parameter description for Tmin
SENSOR.)– Unsuitable download carried out.

– Check configuration and perform
download again.

– Loose connection at sensor cable– Wait a short period of time andtighten the connection, or avoid
loose connection.

33

721 (A721) AlarmB

B>ZERO POSITION level out ofedit limits– LEVEL MIN or LEVEL MAX has
been changed.

– Perform reset (Code 2710) andcarry out calibration again.

41

722 (A722) Alarm
B

B>EMPTY CALIB. or FULLCALIB. out of edit limits– LEVEL MIN or LEVEL MAX has
been changed.

– Perform reset (Code 2710) andcarry out calibration again.

42725 (A725) AlarmB

B>Sensor connection error,
cycle disturbance– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Block off electromagnetic effects
or eliminate source ofdisturbance.

25– Sensor or main electronics
defective.

– Replace sensor or mainelectronics.

– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Process temperature is outsidepermitted range.

– Check temperature present,
reduce or increase if necessary.

– Sensor defective.

– If the process temperature is
within the permitted range,
replace sensor.

– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Pressure is outside permitted
range.

– Check pressure present, reduce or
increase if necessary.

– Sensor defective.

– If the pressure is within thepermitted range, replace sensor.

726 (E726)

727 (E727)

Error
C
Factory
setting:
Warning

Error
B
Factory
setting:
WarningEndress+Hauser

C>Sensor temperature error overrange

B>Sensor pressure error overrange

31

28

43

Troubleshooting

CodeError type/NA 64

728 (A728) AlarmB

729 (A729) Alarm
B

730 (E730)

Error
C
Factory
setting:
Warning

Cerabar S 4 to 20mA HART

Message/description

Cause

MeasureB>RAM error– Fault in the main electronics.

– Briefly disconnect device from the 2power supply.

– Main electronics defective.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 3
power supply.

– Main electronics defective.

– Replace main electronics.

– Pressure measured value has
undershot the value specified for
the Pmin ALARM WINDOW
parameter.

– Check system/pressure measured 46
value.
– Change value for Pmin ALARM
WINDOW if necessary. (See
also Page 36, parameter
description for PminALARM
WINDOW).

– Loose connection at sensor cable– Wait a short period of time andtighten the connection, or avoid
loose connection.

B>RAM error

C>LRV user limits exceeded

Priority

731 (E731)

Error
C
Factory
setting:
Warning

C>URV user limits exceeded– Pressure measured value has
overshot the value specified for
the Pmax ALARM WINDOW
parameter.

– Check system/pressure measured 45
value.
– Change value for Pmax ALARM
WINDOW if necessary. (See
also Page 36, parameter
description for PmaxALARM
WINDOW).

732 (E732)

Error
C
Factory
setting:
Warning

C>LRV Temp. user limits
exceeded– Temperature measured value has
undershot the value specified for
the Tmin ALARM WINDOW
parameter.

– Check system/temperature
measured value.
– Change value for Tmin ALARM
WINDOW if necessary. (See
also Page 37, parameter
description for Tmin PROCESS.)– Loose connection at sensor cable– Wait a short period of time andtighten the connection, or avoid
loose connection.

C>URV Temp. User limits
exceeded– Temperature measured value has
overshot the value specified for
the Tmax ALARM WINDOW
parameter.

– Check system/temperature
measured value.
– Change value for Tmax ALARM
WINDOW if necessary. (See
also Page 37, parameter
description for Tmax PROCESS.)

B>RAM error– Fault in the main electronics.

– Briefly disconnect device from the 4
power supply.

– Main electronics defective.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 20
power supply.

– Main electronics defective.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 19
power supply.

– Main electronics defective.

– Replace main electronics.

– Fault in the main electronics.

– Briefly disconnect device from the 23
power supply.

– Main electronics defective.

– Replace main electronics.

– LEVEL MIN or LEVEL MAX has
been changed.

– Perform reset (Code 2710) andcarry out calibration again.

733 (E733)

Error
C
Factory
setting:
Warning

736 (A736) Alarm
B

737 (A737) AlarmB

738 (A738) AlarmB

739 (A739) AlarmB

741 (A741) AlarmB

44

B>Measurement error

B>Measurement error

B>Measurement error

B>TANK HEIGHT out of edit
limits48

47

44

Endress+Hauser

Cerabar S 4 to 20mA HART

Code

Error type/NA 64

742 (A742) AlarmB

743 (E743)

Alarm
B

744 (A744) Alarm
B

Troubleshooting

Message/description

Cause

MeasurePriority

B>Sensor connection error(upload)– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P. This message
normally only appears briefly.

– Wait a few minutes.
– Perform reset (Code 7864) andcarry out calibration again.

18– Cable connection sensor –main
electronics disconnected.

– Check cable connection and repair
if necessary.

– Sensor defective.

– Replace sensor.

– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P. This message
normally only appears briefly.

– Wait a few minutes.
– Restart the device. Perform reset(Code 62).

– Main electronics defective.

– Replace main electronics.

– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Restart the device. Perform reset(Code 62).
– Block off electromagnetic effects
or eliminate source ofdisturbance.

– Main electronics defective.

– Replace main electronics.

B>Electronic PCB error duringinitialization

B>Main electronic PCB error

14

11

745
(W745)

WarningC

C>Sensor data unknown– Sensor does not suit the device(electronic sensor nameplate).
Device continues measuring.

– Replace sensor with a suitablesensor.

56

746(W746)

WarningC

C>Sensor connection error initializing– Electromagnetic effects are
greater than specifications in thetechnical data. This message
normally only appears briefly.
See Technical Information
TI00383P.

– Wait a few minutes.
– Restart the device. Perform reset(Code 62).
– Block off electromagnetic effects
or eliminate source ofdisturbance.

26– Overpressure or low pressure
present.

– Reduce or increase pressure.

747 (A747) AlarmB

B>Sensor software notcompatible to electronics– Sensor does not suit the device(electronic sensor nameplate).

– Replace sensor with a suitablesensor.

16

748 (A748) Alarm
B

B>Memory failure in signal
processor– Electromagnetic effects are
greater than specifications in thetechnical data.
See Technical Information
TI00383P.

– Block off electromagnetic effects
or eliminate source ofdisturbance.

15– Main electronics defective.

– Replace main electronics.

6.1.1

Onsite display error messages

If the device detects a defect in the onsite display during initialization, the following error
messages can be displayed:

Endress+Hauser

45Troubleshooting

Cerabar S 4 to 20mA HART

Message

MeasureInitialization, VU Electr. Defect
A110

Exchange onsite display.

Initialization, VU Electr. Defect
A114
Initialization, VU Electr. Defect
A281
Initialization, VU Checksum Err.
A110
Initialization, VU Checksum Err.
A112
Initialization, VU Checksum Err.
A171

6.2

Response of outputs to errors

The device makes a distinction between the error types Alarm, Warning and Error.
 See also Section 6.1 "Messages" and Page 21 ff, Table 7: OUTPUT and Page 35 ff,
Table 15: MESSAGES.
OutputA (Alarm)

Current output– Device does not continue measuring. Device continues measuring.
– The current output assumes the value
specified via the OUTPUT FAIL
MODE1), ALT. CURR. OUTPUT1 andSET MAX. ALARM.1
See also Operating Instructions
BA00412P, Section "Configuring
current output for an alarm".

For this error, you can enter whether thedevice should react as in the event of analarm or as in the event of a warning.
See corresponding "Alarm" or "Warning"column.
(See also these Operating
Instructions, parameter description for
SELECT ALARMTYPE.)Bar graph(onsite display)

The bar graph adopts the value definedby the OUTPUT FAIL MODE 1
parameter.

See this table, "Alarm" or "Warning"column, depending on the option
selected.

Onsite display– The measured value and message are – The measured value and message aredisplayed alternately
displayed alternately-symbol– Measured value display:-symbol is – Measured value display:
flashes.
permanently displayed.

Remote operation(digitalcommunication)

W (Warning)

The bar graph adopts the value whichcorresponds to the current value.

– The measured value and message aredisplayed alternately– Measured value display: seecorresponding "Alarm" or "Warning"columnMessage display– 3-digit number such as A122 anddescription

Message display:– 3-digit number such as W613 anddescription

Message display:– 3-digit number such as E731 anddescription

In the case of an alarm, the ALARM
STATUS2) parameter displays a 3-digit
number such as 122 for "Sensor notconnected".

In the case of a warning, the ALARM
STATUS2 parameter displays a 3-digit
number such as 613 for "Simulation is
active".

In the case of an error, the ALARM
STATUS2 parameter displays a 3-digit
number such as 731 for "URV user limits
exceeded".

1)

Menu path: (GROUP SELECTION ) OPERATING MENU OUTPUT

2)

Menu path: (GROUP SELECTION ) OPERATING MENU MESSAGES

46

E (Error: Alarm/Warning)

Endress+Hauser

Cerabar S 4 to 20mA HART

Troubleshooting

6.3

Confirming messages

Depending on the settings for the ALARM DISPL. TIME (Page 36) and ACK. ALARM
MODE (Page 35), the following measures should be taken to clear a message:
Settings1)

Measures– ALARM DISPL. TIME = 0 s– ACK. ALARM MODE = Off– Rectify the cause of the message (see also Section 6.1).

– ALARM DISPL. TIME > 0 s– ACK. ALARM MODE = Off– Rectify the cause of the message (see also Section 6.1).
– Wait for the alarm display time to elapse.

– ALARM DISPL. TIME = 0 s– ACK. ALARM MODE = On

– Rectify the cause of the message (see also Section 6.1).
– Confirm message using ACK. ALARM parameter.

– ALARM DISPL. TIME > 0 s– ACK. ALARM MODE = On

– Rectify the cause of the message (see also Section 6.1).
– Confirm message using ACK. ALARM parameter.
– Wait for the alarm display time to elapse. If a message appears and thealarm display time elapses before the message has been acknowledged,
the message will be cleared once it has been acknowledged.

1)

Menu path for ALARM DISPL. TIME and ACK. ALARM MODE: (GROUP SELECTION ) OPERATING MENUDIAGNOSTICS MESSAGES

If the onsite display displays a message, you can suppress it with the F key.
If there are several messages, the onsite display shows the message which has the highest
priority (Section 6.1). Once you have suppressed this message using the F key, the message
with the next highest priority is displayed. You can use the F key to suppress each message,
one after the other.
The ALARM STATUS parameter continues to display all the messages present.

Endress+Hauser

47

Appendix

Cerabar S 4 to 20mA HART

7

Appendix

7.1

Operating menu for onsite display and digitalcommunication• The entire menu is depicted on the following pages.
• The menu has a different structure depending on the measuring mode selected. Thismeans that some function groups are only displayed for one measuring mode.
• In addition, there are also parameters that are only displayed if other parameters areappropriately configured. For example the CUSTOMER UNIT P parameter is only displayed
if the "User unit" option was selected for the PRESS. ENG. UNIT parameter. These
parameters are indicated with a "*".
• For a description of the parameters, see Section 5 "Description of parameters". The exact
dependency of individual parameters on one another is explained here.

48

Endress+Hauser

Cerabar S 4 to 20mA HART

Appendix

Measured value

1)
GROUP SELECTION

1)LANGUAGE

1)
OPERATING MENUQUICK SETUPLEVEL SELECTION

1)MEASURING MODE

SETTINGSPOSITION ADJUSTMENT
BASIC SETUP

2)2)

MEASURING MODE

MEASURING MODE

Pressure

Pressure
POS. INPUT VALUEPRESS. ENG. UNIT

Level (not provided
for PMP71 with MID)

SET LRVSET URVDAMPING VALUECUSTOMER UNIT P*

CUST. UNIT FACT. P*

POS. INPUT VALUE

SET LRVCALIB. OFFSETSET URV

Level (not provided
for PMP71 with MID)

GET LRVGET URVDAMPING VALUE

1) Display via on-site display only
2) Display via FieldCare and
HART Handheld terminal only

are parameters that are only displayed if other parameters are* Thereappropriately configured.
For example the CUSTOMER UNIT P parameter is only displayed
if the "User unit" option was selected for the PRESS. ENG. UNIT parameter.
These parameters are indicated with a "*".

P01-MIDxxxxx-19-xx-xx-EN-025

Endress+Hauser

49

Appendix

Cerabar S 4 to 20mA HART**

EXTENDED SETUPDISPLAYOUTPUT

SAFETY LOCKMENU DESCRIPTOROUTPUT CURRENTSAFETY PASSWORD

MAIN DATA FORMATCURR. CHARACT.

DIGIT SETS

ALTERNATE DATAOUTPUT FAIL MODE

SAFETY CONFIRM.

Pressure
TEMP. ENG UNIT

Level (not provided
for PMP71 with MID)

2)
CURRENT OUTPUTLANGUAGE

ALT. CURR. OUTPUT

ACK. ALARM MODEDISPLAY CONTRAST

SET MAX. ALARM

CALIB. OFFSETDIGIT SETS

SET MIN. CURRENT

3)ASSIGN CURRENT

MEASURING MODE

Pressure

SET LRV

Level (not provided
for PMP71 with MID)

SET URVDAMPING VALUE

CONF. PASSWORD

2) Display via
HART handheld terminal only

are parameters that are only displayed if other parameters are* Thereappropriately configured.
These parameters are indicated with a "*".

** See Safety Manual SD00190P.
P01-MIDxxxxx-19-xx-xx-en-032

50

Endress+Hauser

Cerabar S 4 to 20mA HART

Appendix

TRANSMITTER INFOPROCESS INFO

HART DATATRANSMITTER DATAPROCESS CONNECTIONSENSOR DATAPROCESS VALUE

BUS ADDRESSDEVICE SERIAL No

Pmax PROC. CONN.

SENSOR SER. NO.

MEASURED VALUEDEVICE TYPE

ELECTR. SERIAL No

PROC. CONN. TYPE

PRESS. SENS LOLIMPRESSURE

MAX. MEAS. PRESS.

CUST. TAG NUMBER

MAT. PROC. CONN. +PRESS. SENS HILIM

CORRECTED PRESS.

COUNTER:P < Pmin

LONG TAG NUMBER

MAT. PROC. CONN. –MINIMUM SPANSENSOR PRESSURE

MIN. MEAS. PRESS.

ADDITIONAL INFO.

SEAL TYPESENSOR MEAS. TYPESENSOR TEMP.

COUNTER:T > Tmax

Pmin SENS. DAMAGE

MEAS. VAL. TREND

MAX. MEAS. TEMP.

PEAK HOLD INDICATOR

Pressure

DEVICE REVISION

Level (not provided
for PMP71 with MID)

COUNTER:P > Pmax

2)
BURST MODE

2)
BURST OPTION

2)PREAMBLE NUMBER

DEVICE DESIGN.

BOLTS MATERIAL

2)
MANUFACTOR ID

HARDWARE REV.

NUTS MATERIAL

Pmax SENS. DAMAGE

COUNTER:T< Tmin

MAT. MEMBRANE

MIN. MEAS. TEMP.

FILLING FLUIDPCB COUNT:T > Tmax

Tmin SENSOR

PCB MAX. TEMP.

Tmax SENSORPCB COUNT: T< Tmin

SENSOR H/WARE REV.

PCB MIN. TEMP.

2)
HART MESSAGE

SOFTWARE VERSIONDRAIN VENT MAT.

2)
HART DATECONFIG RECORDERDRAIN VENT POS.

2)2)PRIMARY VALUE IS

PCB TEMPERATURE

THREAD

2)

2)
PRIMARY VALUEALLOWED MIN. TEMPMOUNTING THREAD

2)

2)
SECONDARY VAL. IS

ALLOWED MAX. TEMPREMOTE SEAL +2)

2)
SECONDARY VALUEDIP STATUS

REMOTE SEAL –THIRD VALUE ISRESET PEAKHOLD

2)2)
DIAPHRAGM MAT. +2)

2)
THIRD VALUE

DIAPHRAGM MAT. –2)

2)
4TH VALUE IS

NR OF REMOTE SEAL

2)

2)
4 TH VALUEFILL FLUID

2) Display via FieldCare and
HART handheld terminal only

are parameters that are only displayed if other parameters are* Thereappropriately configured.
These parameters are indicated with a "*".

P01-MIDxxxxx-19-xx-xx-en-033

Endress+Hauser

51

Appendix

Cerabar S 4 to 20mA HART

OPERATIONSERVICEDIAGNOSTICSSIMULATIONMESSAGES

USER LIMITSSYSTEM 2

ENTER RESET CODESIMULATION MODE

ALARM STATUS

Pmin ALARM WINDOW

CURR. TRIM 4mA

OPERATING HOURSILAST DIAG. CODE

Pmax ALARM WINDOW

CURR. TRIM 20mA

INSERT PIN NO

ACK. ALARM MODE

Tmin ALARM WINDOWOFFSET 4mA TRIM

HistoROM AVAIL.

ACK. ALARM* Tmax ALARM WINDOWOFFSET 20mA TRIM*RESET ALL ALARMS*ERROR NO.

DOWNLOAD SELECT.

HistoROM CONTROL

Display via HART handheld
terminal only

SELECT ALARMTYPE

ALARM DELAY

ALARM DISPL. TIME

I
none

PressureSIM. PRESSURE

Level
SIM. LEVEL

Tank content
SIM. TANK CONT.

Current

SIM. CURRENT

Alarm Warning
SIM. ERROR NO.

There are parameters that are only displayed if other parameters are* appropriately
configured.
These parameters are indicated with a "*".

P01-PMx7xxxx-19-xx-xx-xx-044

52

Endress+Hauser

Cerabar S 4 to 20mA HART

Endress+Hauser

Appendix53

Appendix

54

Cerabar S 4 to 20mA HART

Endress+Hauser

Cerabar S 4 to 20mA HART

Index
Numerics

4TH VALUE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
4TH VALUE IS . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26

A

ACK. ALARM (500). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
ACK. ALARM MODE (401) . . . . . . . . . . . . . . . . . . . . . . . 35
ACK. ALARM MODE (844), "Safety confirm." group . . . . . 6
ADDITIONAL INFO. (272) . . . . . . . . . . . . . . . . . . . . . . . . 26
ALARM DELAY (336) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
ALARM DISPL. TIME (480) . . . . . . . . . . . . . . . . . . . . . . . 36
ALARM STATUS (046) . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
ALLOWED MAX. TEMP (359). . . . . . . . . . . . . . . . . . . . . 26
ALLOWED MIN. TEMP (358) . . . . . . . . . . . . . . . . . . . . . 26
ALT.CURR.OUTPUT (597) . . . . . . . . . . . . . . . . . . . . . . . . 22
ALTERNATE DATA (423) . . . . . . . . . . . . . . . . . . . . . . . . 19

B

BOLTS MATERIAL . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
BURST MODE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
BURST OPTION . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
BUS ADDRESS (345) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

C

CALIB. OFFSET (319) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
CALIB. OFFSET (847), "Safety confirm." group. . . . . . . . . . 6
CONF. PASSWORD (856) . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
CONFIG RECORDER (352) . . . . . . . . . . . . . . . . . . . . . . . . 26
CORRECTED PRESS. (434), "Pressure" measuring mode 31
CURR. CHARACT. (694), (695), (696), (764) . . . . . . . . 22
CURR. TRIM 20mA (042). . . . . . . . . . . . . . . . . . . . . . . . . 38
CURR. TRIM 4mA (045) . . . . . . . . . . . . . . . . . . . . . . . . . . 38
CURRENT OUTPUT (875). . . . . . . . . . . . . . . . . . . . . . . . . . . 6
CUST. TAG NUMBER (055) . . . . . . . . . . . . . . . . . . . . . . . 26
CUST.UNIT FACT.P (317). . . . . . . . . . . . . . . . . . . . . . . . . 16
CUSTOMER UNIT P (075) . . . . . . . . . . . . . . . . . . . . . . . . 16

D

DAMPING VALUE (247) . . . . . . . . . . . . . . . . . . . . . . 13, 17
DAMPING VALUE (855), "Safety confirm." group . . . . . . . 6
DEVICE DESIGN. (350) . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
DEVICE REVISION (699). . . . . . . . . . . . . . . . . . . . . . . . . . 24
DEVICE SERIAL No (354) . . . . . . . . . . . . . . . . . . . . . . . . . 26
DEVICE TYPE (802), Cerabar S . . . . . . . . . . . . . . . . . . . . 24
DIAPHRAG. MAT. - . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
DIAPHRAG. MAT. + . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
DIGITS SET (841), "Safety confirm." group . . . . . . . . . . . . . 6
DIGITS SET, "Display" group. . . . . . . . . . . . . . . . . . . . . . . . 20
DIP STATUS (363). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27
DISPLAY CONTRAST (339) . . . . . . . . . . . . . . . . . . . . . . . 20
DOWNLOAD SELECT (014) . . . . . . . . . . . . . . . . . . . . . . . 33
DRAIN VENT MAT. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
DRAIN VENT POS. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28

E

ELECTR. SERIAL No (386) . . . . . . . . . . . . . . . . . . . . . . . . 26
ENTER RESET CODE (047). . . . . . . . . . . . . . . . . . . . . . . . 32
Endress+Hauser

ERROR No. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35

F

FILL FLUID . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
FILLING FLUID (366) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29

G

GET LRV (309), "Pressure" measuring mode . . . . . . . . . . 17
GET URV (310), "Pressure" measuring mode . . . . . . . . . . 17

H

HARDWARE REV. (266) . . . . . . . . . . . . . . . . . . . . . . . . . . 26
HART DATE (481) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
HART MESSAGE (271). . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
HistoROM AVAIL. (831) . . . . . . . . . . . . . . . . . . . . . . . . . . 33
HistoROM CONTROL (832) . . . . . . . . . . . . . . . . . . . . . . . . 33

I

INSERT PIN NO (048). . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33

L

LANGUAGE (079). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
LANGUAGE, display group. . . . . . . . . . . . . . . . . . . . . . . . . 20
LAST DIAG. CODE (564) . . . . . . . . . . . . . . . . . . . . . . . . . . 35
LONG TAG NUMBER (305) . . . . . . . . . . . . . . . . . . . . . . . . 26

M

MAIN DATA FORMAT (688) . . . . . . . . . . . . . . . . . . . . . . 19
MANUFACTOR ID (432) . . . . . . . . . . . . . . . . . . . . . . . . . . 25
MAT. MEMBRANE (365) . . . . . . . . . . . . . . . . . . . . . . . . . 29
MAT. PROC. CONN. + (360) . . . . . . . . . . . . . . . . . . . . . . . 27
MEAS. VAL. TREND (378). . . . . . . . . . . . . . . . . . . . . . . . . 31
MEASURED VALUE, "Pressure" measuring mode . . . . . . 30
MEASURING MODE (845), "Safety confirm." group . . . . . 6
MEASURING MODE, digital communication . . . . . . 12, 15
MENU DESCRIPTOR (419). . . . . . . . . . . . . . . . . . . . . . . . . 19
MINIMUM SPAN (591) . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
MOUNTING THREAD . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28

N

NR OF REMOTE SE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
NUTS MATERIAL . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28

O

OFFSET 20mA TRIM (044) . . . . . . . . . . . . . . . . . . . . . . . . 38
OFFSET 4mA TRIM (043) . . . . . . . . . . . . . . . . . . . . . . . . . 38
OPERATING HOURS (409). . . . . . . . . . . . . . . . . . . . . . . . . 32
OUTPUT CURRENT (254) . . . . . . . . . . . . . . . . . . . . . . . . . 21
OUTPUT FAIL MODE (388) . . . . . . . . . . . . . . . . . . . . . . . . 22P

PASSWORD (836). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
PCB TEMPERATURE (357) . . . . . . . . . . . . . . . . . . . . . . . . 26
Pmax ALARM WINDOW (333). . . . . . . . . . . . . . . . . . . . . 36
Pmax PROC. CONN. (570) . . . . . . . . . . . . . . . . . . . . . . . . . 27
Pmin ALARM WINDOW (332) . . . . . . . . . . . . . . . . . . . . . 36
POS. INPUT VALUE (563) . . . . . . . . . . . . . . . . . . . . . 13–14

55

Cerabar S 4 to 20mA HART

PREAMBLE NUMBER (036) . . . . . . . . . . . . . . . . . . . . . . . 25
PRESS. ENG. UNIT (060) . . . . . . . . . . . . . . . . . . . . . . . . . . 16
PRESS. SENS HILIM (485) . . . . . . . . . . . . . . . . . . . . . . . . . 29
PRESS. SENS LOLIM (484) . . . . . . . . . . . . . . . . . . . . . . . . 29
PRESSURE, "Pressure" measuring mode . . . . . . . . . . . . . . 31
PRIMARY VALUE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
PRIMARY VALUE IS . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
PROC. CONN. TYPE (482) . . . . . . . . . . . . . . . . . . . . . . . . . 27

Q

Quick Setup "Pressure" menu . . . . . . . . . . . . . . . . . . . . . . . 12

R

REMOTE SEAL - . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
REMOTE SEAL + . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
RESET ALL ALARMS (603). . . . . . . . . . . . . . . . . . . . . . . . 35

S

SAFETY LOCKSTATE (836). . . . . . . . . . . . . . . . . . . . . . . . . 6
SEAL TYPE (362) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
SECONDARY VAL. IS . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
SECONDARY VALUE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
SELECT ALARMTYPE (595), (600) . . . . . . . . . . . . . . . . . 36
SENS H/WARE REV (487) . . . . . . . . . . . . . . . . . . . . . . . . 29
SENSOR MEAS.TYPE (581). . . . . . . . . . . . . . . . . . . . . . . . 29
SENSOR PRESSURE (584), "Pressure" measuring mode . 31
SENSOR SER. No. (250) . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
SENSOR TEMP. (367) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 31
SET LRV (245), "Pressure" measuring mode . . . . . . 13, 16
SET LRV (852), "Safety confirm." group . . . . . . . . . . . . . . . 6
SET MAX. ALARM (342). . . . . . . . . . . . . . . . . . . . . . . . . . 23
SET MIN. CURRENT (343). . . . . . . . . . . . . . . . . . . . . . . . . 23
SET URV (246), "Pressure" measuring mode . . . . . . 13, 16
SET URV (853), "Safety confirm." group . . . . . . . . . . . . . . . 6
SIM. CURRENT (270) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 34
SIM. ERROR NO. (476). . . . . . . . . . . . . . . . . . . . . . . . . . . . 34
SIM. PRESSURE (414) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 34
SIMULATION MODE (413) . . . . . . . . . . . . . . . . . . . . . . . . 34
SOFTWARE VERSION (264) . . . . . . . . . . . . . . . . . . . . . . . 26

T

TEMP. ENG. UNIT (318), "Pressure" measuring mode . . 18
THIRD VALUE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
THIRD VALUE IS . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
THREAD PROCESS . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
Tmax ALARM WINDOW (335) . . . . . . . . . . . . . . . . . . . . 37
Tmax SENSOR (369) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Tmin ALARM WINDOW (334). . . . . . . . . . . . . . . . . . . . . 37
Tmin SENSOR (368). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29

56

Endress+Hauser

Cerabar S 4 to 20mA HART

Endress+Hauser

57

58

59

71335863

www.addresses.endress.com