Environmental Sensor Packages
Guide

Revision 0B

Copyright © 2015 Raritan, Inc.
SENSOR-0B-v3.1-E
March 2015
255-90-0003-00

This document contains proprietary information that is protected by copyright. All rights reserved. Nopart of this document may be photocopied, reproduced, or translated into another language without
express prior written consent of Raritan, Inc.
© Copyright 2015 Raritan, Inc. All third-party software and hardware mentioned in this document areregistered trademarks or trademarks of and are the property of their respective holders.
FCC Information

This equipment has been tested and found to comply with the limits for a Class A digital device,pursuant to Part 15 of the FCC Rules. These limits are designed to provide reasonable protection
against harmful interference in a commercial installation. This equipment generates, uses, and canradiate radio frequency energy and if not installed and used in accordance with the instructions, maycause harmful interference to radio communications. Operation of this equipment in a residential
environment may cause harmful interference.
VCCI Information (Japan)

Raritan is not responsible for damage to this product resulting from accident, disaster, misuse, abuse,non-Raritan modification of the product, or other events outside of Raritan's reasonable control or notarising under normal operating conditions.
If a power cable is included with this product, it must be used exclusively for this product.

Contents
Chapter 1

Introduction to Environmental Sensor Packages

1

Sensor Overview............................................................................................................................ 2
Sensor Comparison ....................................................................................................................... 2

Chapter 2

DPX Series

4

Available DPX Sensor Packages................................................................................................... 5
DPX Temperature and Humidity Sensors...................................................................................... 6
Air Flow Sensors............................................................................................................................ 6
Differential Air Pressure Sensors................................................................................................... 7
Connecting Tubes ...............................................................................................................8
Cascading Air Pressure Sensors......................................................................................... 8Contact Closure Sensors............................................................................................................... 9
Old and New Contact Closure Sensors............................................................................. 11
Connecting Detectors/Switches to New DPX-CC2-TR ..................................................... 11
Connecting Detectors/Switches to Old DPX-CC2-TR ....................................................... 13
Configuring a Contact Closure Sensor.............................................................................. 14
Contact Closure Sensor LEDs........................................................................................... 16
Floor Water Sensors .................................................................................................................... 17
Floor-mounted Water Sensor ............................................................................................ 17
Cable Water Sensor .......................................................................................................... 18

Chapter 3

DPX2 Series

20

Available DPX2 Sensor Packages............................................................................................... 20
DPX2 Temperature and Humidity Sensors.................................................................................. 21
DPX2 LED States .............................................................................................................. 21
Connection and Disconnection .................................................................................................... 22
Connecting a DPX2 Sensor Cable .................................................................................... 23
Disconnecting a DPX2 Sensor .......................................................................................... 24iii

Contents

Chapter 4

DX Series

25

Available DX Sensor Packages ...................................................................................................26
DX-D2C6...................................................................................................................................... 27
DX-PD2C5 ................................................................................................................................... 30
DX-PIR......................................................................................................................................... 33
DX-D4C3...................................................................................................................................... 35
DX-VBR ....................................................................................................................................... 38
Making Connections .................................................................................................................... 40
Pre-installed DX Brackets.................................................................................................. 40
Connecting Detectors/Actuators to DX.............................................................................. 40
Cascading DX Sensor Packages ...................................................................................... 43
Connecting a DPX2 Sensor Package to DX ..................................................................... 44
Adjusting Dip Switches ................................................................................................................ 45

Appendix A

Supported Maximum Sensor Distance46

Appendix B

Supported Maximum Number of Sensors and Actuators

47

Appendix C

Sensor Measurement Accuracy

49

DPX Sensor Accuracy ................................................................................................................. 49
DPX2 Sensor Accuracy ............................................................................................................... 49

Appendix D

Sensor Firmware Update

50

Appendix E

Sensor Naming Conventions

51

Index

iv

53

Chapter 1

Introduction to Environmental
Sensor Packages
Raritan has three types of environmental sensor packages: DPX, DPX2
and DX series.
DPX series described in this guide refers to DPX sensor packages
instead of DPX power distribution units (PDUs) unless otherwise
specified.
Note that NOT all Raritan products support all types of Raritan
environmental sensor packages. For specific sensor support information,
see the online help of your Raritan EMX or PDU like PX2, PX3 and so on.
Or see Supported Maximum Number of Sensors and Actuators (onpage 47) for a list of Raritan products that support these environmental
sensor packages.
Note: Online help is accessible from your Raritan PDU or EMX
application, or the Support page (http://www.raritan.com/support/) on
Raritan's website.

In This Chapter
Sensor Overview .......................................................................................2
Sensor Comparison...................................................................................2

1

Chapter 1: Introduction to Environmental Sensor Packages

Sensor Overview
An environmental sensor package may have more than one sensor or
actuator function. For details, see:

Available DPX Sensor Packages (on page 5)Available DPX2 Sensor Packages (on page 20)Available DX Sensor Packages (on page 26)

A sensor is used to detect the environmental condition, such astemperature, humidity, and water presence. An actuator is used tocontrol a system or mechanism, such as turning on or off a device.
Depending on the firmware of your Raritan PDU or EMX, the maximum
number of sensors or actuators supported by your PDU or EMX differs.
See Supported Maximum Number of Sensors and Actuators (onpage 47).
The maximum cabling length for all sensor packages connected to a
Raritan product's sensor port should not exceed 98 feet (30 meters). See
Supported Maximum Sensor Distance (on page 46).
Exception: DX-PD2C5 supports a maximum cabling length of 29 feet (9m) instead of 98 feet (30 m).
Warning: For proper operation, wait for 15~30 seconds between eachconnection operation or each disconnection operation ofenvironmental sensor packages.

Sensor Comparison

2Chapter 1: Introduction to Environmental Sensor Packages

Sensorfamily

DPX
DPX2
DX

Connection
interface
RJ-12
RJ-12

Support for
sensor daisy
chain

Chain position
availability*

Support for a
Raritan sensor
hub

Automatic
sensor firmware
update

**
***

RJ-45
* A sensor's position in a sensor daisy chain is included in the chain
position availability information. This information can be retrieved throughthe Raritan PDU or EMX where the sensor is connected.
** Most DPX sensor packages do NOT support daisy chaining except for
DPX-T1DP1.
*** DPX2 sensors can be daisy chained via one DPX2 sensor cable, but
you cannot daisy chain multiple DPX2 sensor cables.

3

Chapter 2

DPX Series
DPX is the first generation of Raritan environmental sensor packages.
Most DPX packages come with a factory-installed sensor cable with an
RJ-12 connector.
DPX supports the use of a Raritan sensor hub to increase the number ofconnected DPX sensors.

In This Chapter
Available DPX Sensor Packages ..............................................................5
DPX Temperature and Humidity Sensors .................................................6
Air Flow Sensors .......................................................................................6
Differential Air Pressure Sensors ..............................................................7
Contact Closure Sensors...........................................................................9
Floor Water Sensors................................................................................17

4

Chapter 2: DPX Series

Available DPX Sensor Packages

Sensor packages
DPX-T1

DPX-T1H1

Description
 1 temperature sensor
See DPX Temperature and Humidity Sensors(on page 6).


1 temperature sensor



1 humidity sensor

See DPX Temperature and Humidity Sensors(on page 6).
DPX-T3H1



3 temperature sensors



1 humidity sensor

See DPX Temperature and Humidity Sensors(on page 6).
DPX-AF1

DPX-T1DP1

1 air flow sensor

See Air Flow Sensors (on page 6).


1 temperature sensor



1 differential air pressure sensor

See Differential Air Pressure Sensors (on page
7).
DPX-CC2-TR

DPX-WSF

DPX-WSC series

Two-channel contact closure sensors

See Contact Closure Sensors (on page 9).


1 floor-mounted water sensor

See Floor-mounted Water Sensor (on page 17).


1 cable water sensor

See Cable Water Sensor (on page 18).

For detailed information on sensor names, see Sensor Naming
Conventions (on page 51).

5

Chapter 2: DPX Series

DPX Temperature and Humidity Sensors
Raritan provides three types of DPX temperature and humidity sensor
packages: DPX-T1, DPX-T1H1 and DPX-T3H1.
A DPX-T3H1 package contains one DPX-T1H1 and two DPX-T1 sensors.
As shown in the following diagram, all sensors have been connected tothe sensor cable, when shipping out of the factory, in a manner that you
cannot remove or replace any individual sensor.

Air Flow Sensors
If a DPX air flow sensor (DPX-AF1) is connected, make sure the sensor
faces the source of the wind (such as a fan) in the appropriate orientation
as indicated by the arrow on that sensor.

To affix this sensor to an object or place, just screw it up using thesensor's two screw holes.

6

Chapter 2: DPX Series

Differential Air Pressure Sensors
The DPX differential air pressure sensor (DPX-T1DP1) detects not only
the differential air pressure but also the temperature because it has abuilt-in temperature sensor.
This sensor is designed to receive the inputs of two pressure levels.
Differential air pressure is measured by reading the difference of the two
inputs.
If multiple differential air pressure sensors are needed, you can cascadethem. See Cascading Air Pressure Sensors (on page 8).

NumberComponent

Description'In' port

Connect to a Raritan PDU or EMX.

(RJ-12)

If cascading multiple air pressure sensors,connect this port to the 'Out' port of anothersensor.

'Out' port

Use this port for cascading air pressure
sensors.

(RJ-12)'Lo' pressure
nozzleReceive low air pressure inputs.

'Hi' pressure
nozzleReceive high air pressure inputs.

7Chapter 2: DPX Series

Connecting Tubes
Two tubes are shipped with the sensor. Connect the tubes to the sensor
to receive two air pressure inputs. If necessary, cut the tubes so that the
tube length meets your needs.
To install tubes:
1. Connect the tubes to both pressure nozzles of the sensor.

2. Place each tube's grommet in an appropriate location where youwant to detect differential air pressure levels.
Place the "Lo" tube's grommet in a low air pressure position.

Place the "Hi" tube's grommet in a high air pressure position.

Cascading Air Pressure Sensors
You can increase the number of differential air pressure sensors by
cascading them. Make sure the number of cascaded air pressure
sensors does not exceed the maximum number of sensors supported byyour Raritan PDU or EMX. The supported maximum number varies
based on the firmware version. See Supported Maximum Number of
Sensors and Actuators (on page 47).
To cascade multiple sensors:
1. Use a Raritan-provided phone cable to connect an air pressure
sensor to a Raritan PDU or EMX.
Plug one end of the cable into the sensor's "In" port.



Plug the other end into a Raritan PDU's or EMX' sensor port.
An RJ-12 to RJ-45 adapter is required if your Raritan PDU uses
an RJ-45 SENSOR port, such as PX3.

8

Chapter 2: DPX Series

2. Use another Raritan-provided phone cable to connect the first airpressure sensor to an additional one.
a. Plug one end of the cable into the "Out" port of the first sensor.
b. Plug the other end into the "In" port of the additional sensor.
3. Repeat the above step if intending to cascade more air pressure
sensors.

Contact Closure Sensors
Raritan's contact closure sensor (DPX-CC2-TR) can detect the
open-and-closed status of connected detectors/switches.
This sensor has two channels for connecting two discrete (on/off)detectors/switches. Four termination points are available: the two to theright are associated with one channel (as indicated by the sensor's LED
number), and the two to the left are associated with the other.

NumberComponent
Two LEDs for indicating two channels' status.
Four termination points for connecting two discrete
detectors/switches.

9Chapter 2: DPX Series

Supported detector/switch types:
At a minimum, a discrete detector/switch is required in order for
DPX-CC2-TR to work properly. The types of discrete detectors/switches
that can be connected to DPX-CC2-TR include those for:

Door open/closed detection



Door lock detection

Floor water detection

Smoke detection



Vibration detection

Raritan does NOT produce most of the above detectors/switches except
floor water sensors. When using third-party probes, you must test themwith DPX-CC2-TR to ensure they work properly.
Important: Integration and testing for third-party detectors/switches is thesole responsibility of the customer. Raritan cannot assume any liability
as a result of improper termination or failure (incidental or consequential)of third-party detectors/switches that customers provide and install.
Failure to follow installation and configuration instructions can result infalse alarms or no alarms. Raritan makes no statement or claim that all
third-party detectors/switches will work with DPX-CC2-TR.
Connection scheme:
A DPX-CC2-TR can connect two contact closure detectors/switches.

For details on connecting a detector/switch, see Connecting
Detectors/Switches to New DPX-CC2-TR (on page 11) or Connecting
Detectors/Switches to Old DPX-CC2-TR (on page 13).

10

Chapter 2: DPX Series

Old and New Contact Closure Sensors
Since the end of 2012 Raritan provided a new contact closure sensor.
Old and new contact closure sensors are functionally identical with thefollowing exceptions.


The old sensor type uses buttons to secure or loosen the connectionof detectors/switches while the new type uses screws.

The new sensor type's terminals can be separated from the sensor
but the old type's terminals are fixed.

The new sensor type supports Raritan's DPX water sensors(DPX-WSC and DPX-WSF) but the old type does NOT.

Both old and new contact closure sensors share the same part number:
DPX-CC2-TR.
The latest firmware for your Raritan PDU or EMX will continuously
support the old DPX-CC2-TR.
For instructions on new contact closure sensors, see Connecting
Detectors/Switches to New DPX-CC2-TR (on page 11).

For instructions on old contact closure sensors, see Connecting
Detectors/Switches to Old DPX-CC2-TR (on page 13).

Connecting Detectors/Switches to New DPX-CC2-TR
The new DPX-CC2-TR comprises two parts: sensor box and terminal
module. The terminal module is detachable so it is more convenient toconnect/disconnect discrete detectors/switches.
Below are the resistance values for the new DPX-CC2-TR to open andclose a connected detector/switch reliably.


Open: Greater than 320K ohm



Closed: Less than 200K ohm
To make connections when the terminal module is attached:

1. Strip the insulation around 12 mm from the end of each wire of
discrete detectors/switches.
2. Fully insert each wire of both detectors/switches into eachtermination point.


Plug both wires of a detector/switch into the two termination
points to the left.

11Chapter 2: DPX Series

Plug both wires of the other detector/switch into the twotermination points to the right.

3. Use a screwdriver with a 2.5 mm wide shaft to tighten the screws
above each termination point to secure the wires, using a torque of0.196 N·m (2 kgf·cm).

To make connections after the terminal module is separated:
You can also connect a detector/switch when the terminal module isseparated from the sensor box.
1. Loosen the screws at two sides of the terminal module.

12Chapter 2: DPX Series

Note: The two screws are not removable so just loosen them.

2. Separate the terminal module from the sensor box.

3. After connecting detectors/switches to the terminal module, plug theterminal module back into the sensor box, and then tighten thescrews at two sides of the terminal module.

Connecting Detectors/Switches to Old DPX-CC2-TR
The old DPX-CC2-TR has four spring-loaded termination points and four
little buttons to control the spring of each termination point.
To connect third-party or Raritan's discrete detectors/switches:
1. Strip the insulation around 12 mm from the end of each wire of
discrete detectors/switches.

13

Chapter 2: DPX Series

2. Press and hold down the tiny rectangular buttons above thetermination points on DPX-CC2-TR.

3. Fully insert each wire of both detectors/switches into eachtermination point.


Plug both wires of a detector/switch into the two termination
points to the left.



Plug both wires of the other detector/switch into the twotermination points to the right.

4. Release the tiny rectangular buttons after inserting the wires
properly.
5. Verify that these wires are firmly fastened.
Configuring a Contact Closure Sensor
Before using DPX-CC2-TR to detect contact closure status, water,
smoke or vibration, you must determine the normal state by adjusting its
dip switch. The dip switch controls the LED state on DPX-CC2-TR, andis associated with a channel.
To adjust the dip switch setting:
1. Place the detectors/switches connected to DPX-CC2-TR to theposition where you want to detect a specific environmental situation.
2. Uncover the dip switch on DPX-CC2-TR.


14

New DPX-CC2-TR

Chapter 2: DPX Series

Old DPX-CC2-TR

3. To set the Normal state for channel 1, locate the dip switch labeled
1.
4. Use a pointed tip such as a pen to set the slide switch to the position
labeled N.O or N.C.


N.O (Normally Open): The open status of the connected
detector/switch is considered normal.



N.C (Normally Closed): The closed status of the connected
detector/switch is considered normal.

For Raritan's DPX water sensors, the Normal state must be Normally
Open, which indicates there is no water detected. Set the dip switch
to Normally Open and verify that the LED of the channel where theRaritan's water sensor is connected remains OFF.
New DPX-CC2-TR



Old DPX-CC2-TR

5. To set the Normal state for channel 2, repeat Step 4 for adjusting theother dip switch's setting.

15

Chapter 2: DPX Series

6. Install back the dip switch cover.
Note: The dip switch setting must be properly configured, or the sensor's
LED may be incorrectly lit in the Normal state.
Contact Closure Sensor LEDs
Description in this section applies to LEDs of DPX-CC2-TR and those of
DX contact closure sensor channels. See DX Series (on page 25).
LEDs indicate states of detectors/switches connected to the contact
closure sensor channels.
The LED is lit when the associated detector/switch is in the "alarmed"
state. See Configuring a Contact Closure Sensor (on page 14) for
how to set the Normal state of DPX-CC2-TR.
The meaning of a lit LED varies depending on the Normal state settings.
When the Normal state is set to Normally Closed (N.C):
LED

Sensor state

Off

ClosedLit

Open

When the Normal state is set to Normally Open (N.O):
LED

Sensor state

Off

Open

Lit

ClosedFor Raritan's DPX water sensors, the Normal state must be set toNormally Open (N.O). The following is the correct LED behavior
based on proper dip switch settings.

16

LED

Sensor stateOff

No water detected

Lit

Water detected

Chapter 2: DPX Series

Floor Water Sensors
Raritan has two types of DPX floor water sensors that work in
conjunction with Raritan's new contact closure sensor (DPX-CC2-TR).
One is the floor-mounted water sensor (DPX-WSF), and the other is thecable water sensor (DPX-WSC series).
Note that only the NEW DPX-CC2-TR support these water sensors.
Note: If you order a Raritan DPX water sensor with the part number
containing the suffix -KIT, you get the new DPX-CC2-TR that supports it.
Floor-mounted Water Sensor
The floor-mounted water sensor (DPX-WSF) has a flat bottom so it canstand on the ground. The water detector's dimension is 63.5 mm x 13.25mm x 13.2 mm (W x L x D).
Use one of the following methods to affix this detector to the ground:

Screw the detector to the ground with your own screws. To avoid
breaking the detector's enclosure, do not over tighten the screws.



Put something weighing around 100 to 250 grams (0.22 to 0.55
pounds) into the empty bag shipped with this water sensor. Then use
cable ties to wrap the bag right above the detector. Make sure thedetector is not tilted so that its bottom evenly contacts the ground.

NumberItem
Water detector.

17

Chapter 2: DPX Series

NumberItem
Wires to connect to the contact closure sensor.
You need a minimum of 30 cm wires to prevent thecontact closure sensor from being damaged by floor
water (if any). Below are the wire length limitations:

Minimum: 30 cm (11.8 inches)Maximum: 150 cm (59 inches)

Cable Water Sensor
The cable water sensor is in the shape of a cable so it can be flexibly
placed, twisted or wrapped around a location where water may drip, suchas a ceiling tile, water pipe or the floor.
Raritan provides two types of cable water sensors. The only difference istheir cable length.


DPX-WSC-35: 3.5 meters (11.5 feet).



DPX-WSC-70: 7 meters (23 feet).

NumberItem
Cable-shaped water detector.

18

Chapter 2: DPX Series

NumberItem
Wires to connect to the contact closure sensor.
You need a minimum of 30 cm wires to prevent thecontact closure sensor from being damaged by floor
water (if any). Below are the wire length limitations:

Minimum: 30 cm (11.8 inches)Maximum: 150 cm (59 inches)

19

Chapter 3

DPX2 Series
DPX2 sensors are physically similar to DPX sensors. Below are theirdifferences:

DPX2 does not have a factory-installed sensor cable so you must
manually connect sensors to a DPX2 sensor cable.



DPX2 has an LED to show the sensor status.



DPX2 provides the chain position information.



Individual DPX2 sensors can be replaced without the need toreconnect the DPX2 sensor cable.



DPX2 does NOT support the use of a Raritan sensor hub.

In This Chapter
Available DPX2 Sensor Packages.......................................................... 20
DPX2 Temperature and Humidity Sensors............................................. 21
Connection and Disconnection ............................................................... 22

Available DPX2 Sensor Packages

Sensor packages
DPX2-T1

DPX2-T1H1

Description
 1 temperature sensor
See DPX2 Temperature and Humidity Sensors(on page 21).


1 temperature sensor



1 humidity sensor

See DPX2 Temperature and Humidity Sensors(on page 21).
DPX2-T2H1

2 temperature sensors



1 humidity sensor

See DPX2 Temperature and Humidity Sensors(on page 21).
DPX2-T3H1

3 temperature sensors



1 humidity sensor

See DPX2 Temperature and Humidity Sensors(on page 21).
For detailed information on sensor names, see Sensor Naming
Conventions (on page 51).

20

Chapter 3: DPX2 Series

DPX2 Temperature and Humidity Sensors
There are four types of DPX2 temperature and humidity sensor
packages: DPX2-T1, DPX2-T1H1, DPX2-T2H1, and DPX2-T3H1.
A DPX2-T2H1 package contains one DPX2-T1H1 and one DPX2-T1
sensor.
A DPX2-T3H1 package contains one DPX2-T1H1 and two DPX2-T1
sensors.
Each DPX2 package is shipped with a DPX2 sensor cable for sensor
connection.
The following is a DPX2 sensor's diagram.

NumberComponent

Function

Head connector

Connect to the DPX2 sensor cable.

LED

Indicate the sensor status. See DPX2
LED States (on page 21).

DPX2 LED States
The DPX2 sensor indicates an alert by making its LED flash on and off.
The LED also flashes when the sensor is upgrading its firmware.
DPX2-T1
LED states

Description

Off

Temperature is in the normal state.

Flashing

Temperature enters either alarmed state:warning or critical.

(1 flash and 1 off)
High-speed flashingSensor firmware is being updated. See
Sensor Firmware Update (on page 50).

21

Chapter 3: DPX2 Series

DPX2-T1H1
LED states

Description

Off

Both temperature and humidity are in thenormal state.
1 flash and
1 off

Temperature enters an alarmed state:warning or critical.
Humidity is in the normal state.

Flashing

2 flashesand 1 off

Temperature is in the normal state.

3 flashesand 1 off

Both temperature and humidity enter an
alarmed state: warning or critical.

High-speed flashing

Humidity enters an alarmed state: warningor critical.

Sensor firmware is being updated.

Connection and Disconnection
Before connecting a DPX2 sensor to a Raritan PDU or EMX, you must
connect it to a DPX2 sensor cable first.
If any connected sensor is broken, you can replace it with a new one
without disconnecting the DPX2 sensor cable from the Raritan PDU or
EMX.

22

Chapter 3: DPX2 Series

Connecting a DPX2 Sensor Cable
Raritan has three types of DPX2 sensor cables. The difference is thenumber of available head connectors on the cable: one to three head
connectors.
DPX2 sensor cables cannot be daisy chained.
The following diagram illustrates a DPX2 sensor cable with three head
connectors.

NumberComponent

Function

RJ-12 connector

Connect to a Raritan PDU or EMX.

Head connectors

Connect DPX2 sensors.

Sensor connection guidelines:
Always make sure there are NO free head connectors between thesensor cable's RJ-12 connector and the final DPX2 sensor attached to it.
That is, each head connector prior to the final DPX2 sensor must be
occupied with a sensor. Otherwise, those sensors following the "free"head connector(s) on the same sensor cable do not work properly.
Below illustrates the scenarios when the number of DPX2 sensors is lessthan the number of head connectors on a DPX2 sensor cable.


Scenario A: When only one sensor is connected, always connect itto the first head connector.

23

Chapter 3: DPX2 Series



Scenario B: When only two sensors are connected, always connect
them to the first two head connectors.

Disconnecting a DPX2 Sensor
You can remove any individual DPX2 sensor from the sensor cablewhenever needed.
To disconnect a DPX2 sensor:
1. Press the latch of the cable connector so that the other side of thelatch slightly goes up.

2. Pull the sensor away.
3. Connect any DPX2 sensor back to this free head connector, or move
all subsequent DPX2 sensors that follow it on the same sensor cableto prior head connectors.
Warning: The final step is required, or all subsequent DPX2 sensors
no longer work properly. For example, after removing the sensor fromthe 2nd head connector, you must either add a DPX2 sensor back to
the 2nd head connector, or move the sensor attached to the 3rd head
connector to the 2nd head connector.

24

Chapter 4

DX Series
A DX sensor package consists of terminals, sensors and two RJ-45
ports.
Different DX sensor packages have a different number of terminals.
Every pair of terminals is a channel for connecting a sensor or actuator.
Contact closure channels on DX do not support Raritan's DPX water
sensors.
Note: DX may also not support third-party leakage detectors so you must
test them before using any third-party leakage detectors.
No sensor cables are shipped with DX sensor packages except for an
RJ-12 to RJ-45 adapter cable (part number: 254-01-0032-00). A
standard network patch cable (CAT5e or higher) is needed to:Connect a DX sensor to a Raritan product with an RJ-45 SENSOR
port, such as PX3.
Note: For a Raritan EMX or PDU with an RJ-12 SENSOR port, use
the RJ-12 to RJ-45 adapter cable instead which is shipped with DX.

Daisy chain DX sensors packages.

Warning: Do NOT use a crossover cable to connect DX.

In This Chapter
Available DX Sensor Packages...............................................................26
DX-D2C6 .................................................................................................27
DX-PD2C5 ...............................................................................................30
DX-PIR.....................................................................................................33
DX-D4C3 .................................................................................................35
DX-VBR ...................................................................................................38
Making Connections ................................................................................40
Adjusting Dip Switches ............................................................................45

25

Chapter 4: DX Series

Available DX Sensor Packages
This table lists all DX sensor packages which are available at the time ofwriting.
Sensor packages
DX-D2C6

Description
 7 pairs of terminals:
Two for connecting dry contact signal actuators(DC) and five for contact closure sensors (CC).


1 hall effect sensor

Note: The hall effect sensor is reserved for future
use and currently shall NOT be used.
See DX-D2C6 (on page 27).
DX-PD2C5

7 pairs of terminals:
Two for powered dry contact signal actuatorsand five for contact closure sensors.

See DX-PD2C5 (on page 30).
DX-PIR

1 occupancy sensor



1 tamper sensor



1 pair of terminals for contact closure sensors

See DX-PIR (on page 33).
DX-D4C3

7 pairs of terminals:
Four for connecting dry contact signal actuators(DC) and three for contact closure sensors(CC).

See DX-D4C3 (on page 35).
DX-VBR

1 vibration sensor



1 pair of terminals for contact closure sensors

See DX-VBR (on page 38).
For detailed information on sensor names, see Sensor Naming
Conventions (on page 51).

26

Chapter 4: DX Series

DX-D2C6
DX-D2C6 has seven channels for both contact closure sensors and dry
contact signal actuators.

The label attached to DX-D2C6 helps you identify different channels.
DX-D2C6 Label:

CC represents a contact closure sensor channel. There are five CC
channels: CC1 through CC5.



DC represents a dry contact signal actuator channel. There are two
DC channels: DC1 and DC2.

27

Chapter 4: DX Series

DX-D2C6 hall effect sensor (reserved):
DX-D2C6 contains a built-in hall effect sensor, which can detect whether
a door is open or closed. The hall effect sensor is reserved for future use
and currently shall NOT be used. After connecting DX-D2C6 to a Raritan
PDU or EMX, this built-in sensor will be detected and show up in thatPDU or EMX web interface and SNMP MIB, which is normal.
DX-D2C6 terminals, dip switches, and LEDs:
Terminals, dip switches, and LEDs are located in two rows as shown
below.

Numbers

Components
CC and DC channels.


Top row:
Four CC channels (CC2 - CC5).
Two DC channels (DC1 - DC2).



Bottom row:
One CC channel (CC1).

See Connecting Detectors/Actuators to DX (on page 40)
for how to connect CC sensors or DC actuators.
Dip switches for configuring the Normal state of each CC
channel. See Adjusting Dip Switches (on page 45).
Top row:
Dip switch 1 controls CC2.
Dip switch 2 controls CC3.
Dip switch 3 controls CC4.
Dip switch 4 controls CC5.

28

Chapter 4: DX Series

Numbers

ComponentsBottom row:
Dip switch 1 controls CC1.

Dip switch 2 controls the built-in hall effect sensor.
CC status LEDs. See Contact Closure Sensor LEDs (onpage 16) for details.
High-speed flashing of CC1 LED indicates that the DX
firmware upgrade is in progress. See Sensor Firmware
Update (on page 50).
Top row:
The four LEDs, from left to right, indicate the states ofCC2, CC3, CC4 and CC5 respectively.

Bottom row:
The LED indicates the CC1 state.

29

Chapter 4: DX Series

DX-PD2C5
DX-PD2C5 is physically similar to DX-D2C6 except for the following
differences:

Dry contact signal channels of DX-PD2C5 supply DC 12V power tothe connected actuators.

Dry contact signal channels of DX-PD2C5 only support theconnection of EMKA (1150-U5x) doorhandles.

DX-PD2C5 works with PX3 PDUs and PX3TS transfer switches only.



Only one DX-PD2C5 is supported per PX3 or PX3TS. Daisy chainingof DX-PD2C5 is NOT supported.



DX-PD2C5 supports a maximum cabling length of 29 feet (9 m)
instead of 98 feet (30 m). For details, see Supported Maximum
Sensor Distance (on page 46).

Warning: If high security is required, it is strongly recommended that
DX-PD2C5 shall NOT be used.

A label is attached to DX-PD2C5 to help you identify different channels.
DX-PD2C5 Label:30CC represents a contact closure sensor channel. There are five CC
channels: CC1 through CC5.



PDC represents a dry contact signal actuator channel which is
powered. There are two PDC channels: PDC1 and PDC2. Note thateach PDC channel has two electrical polarity markings below it: (negative) and + (positive), which you must follow when connecting
an EMKA doorhandle.

Chapter 4: DX Series

DX-PD2C5 terminals, dip switches, and LEDs:
Terminals, dip switches, and LEDs are separated into two rows asshown below.

Numbers

Components
CC and PDC channels.


Top row:
Four CC channels (CC2 - CC5).
Two PDC channels (PDC1 - PDC2).



Bottom row:
One CC channel (CC1).

See Connecting Detectors/Actuators to DX (on page 40)
for how to connect CC sensors or DC actuators.
Dip switches for configuring the Normal state of each CC
channel. See Adjusting Dip Switches (on page 45).
Top row:
Dip switch 1 controls CC2.
Dip switch 2 controls CC3.
Dip switch 3 controls CC4.
Dip switch 4 controls CC5.



Bottom row:
Dip switch 1 controls CC1.

31

Chapter 4: DX Series

Numbers

Components

Note: Dip switch 2 in the bottom row does not control any
channel and can be ignored.
CC status LEDs. See Contact Closure Sensor LEDs (onpage 16) for details.
High-speed flashing of CC1 LED indicates that the DX
firmware upgrade is in progress. See Sensor Firmware
Update (on page 50).
Top row:
The four LEDs, from left to right, indicate the states ofCC2, CC3, CC4 and CC5 respectively.

Bottom row:
The LED indicates the CC1 state.

32

Chapter 4: DX Series

DX-PIR
DX-PIR contains one occupancy sensor (that is, presence detector), one
tamper sensor and a pair of terminals for connecting a contact closure(CC) sensor.

Occupancy sensor:
The occupancy sensor is located on the top of the DX-PIR. It uses thepassive infrared technology to detect the motion of a person by sensing
the temperature differences between a person and the surroundings.

The diagrams below show the occupancy sensor's detection range andsensing area.

33

Chapter 4: DX SeriesSide View
 Maximum range: 5 meters

Top View
 Maximum range: 5 meters

 Vertical sensing area: 82 degrees

 Horizontal sensing area: 94 degrees(+/- 41 degrees)(+/- 47 degrees)

Conditions for the detected target:

Temperature difference between the target and the surroundings
should exceed 4 degrees Celsius (7.2 degrees Fahrenheit).



Target's movement speed: 1.0 m/s.



Target concept is human body (size: 700 x 250 mm).
Tamper sensor:

The tamper sensor is located on the back of the DX-PIR. This sensor isused to detect whether the DX-PIR is moved away from its original
position. In the normal state, the DX-PIR is firmly affixed to an object so
this sensor, which is a button on the back of DX-PIR, is pressed down. If
someone takes the DX-PIR away, the button springs up so the sensor isin the alarmed state.

34

Chapter 4: DX Series

DX-PIR Terminals, dip switches and LED:

Numbers

Components
One CC channel comprising a pair of terminals.
See Connecting Detectors/Actuators to DX (on page 40)
for how to connect a CC sensor.
 Dip switch 1 configures the Normal state of the CC
channel.
 Dip switch 2 controls the built-in occupancy sensor.
See Adjusting Dip Switches (on page 45).
CC status LED. See Contact Closure Sensor LEDs (onpage 16) for details.
High-speed flashing of CC1 LED indicates that the DX
firmware upgrade is in progress. See Sensor Firmware
Update (on page 50).

DX-D4C3
DX-D4C3 contains four channels for dry contact signal actuators andthree channels for contact closure sensors.

35

Chapter 4: DX Series

The label attached to DX-D4C3 helps you identify each channel.
DX-D4C3 Label:

CC represents a contact closure sensor channel. There are three CC
channels: CC1 through CC3.



DC represents a dry contact signal actuator channel. There are four
DC channels: DC1 through DC4.
DX-D4C3 terminals, dip switches, and LEDs:

Terminals, dip switches, and LEDs are located in two rows as shown
below.

Numbers

Components
CC and DC channels.


Top row:
2 CC channels (CC2 - CC3).
4 DC channels (DC1 - DC4).



Bottom row:
One CC channel (CC1).

See Connecting Detectors/Actuators to DX (on page 40)
for how to connect CC sensors or DC actuators.
Dip switches for configuring the Normal state of each CC
channel. See Adjusting Dip Switches (on page 45).


36

Top row:Chapter 4: DX Series

Numbers

Components
Dip switch 1 controls CC2.
Dip switch 2 controls CC3.



Bottom row:
Dip switch 1 controls CC1.

Note: Dip switches 3 and 4 in the top row and dip switch 2 inthe bottom row do not control any channel and can beignored.
CC status LEDs. See Contact Closure Sensor LEDs (onpage 16) for details.
High-speed flashing of CC1 LED indicates that the DX
firmware upgrade is in progress. See Sensor Firmware
Update (on page 50).
Top row:
The two leftmost LEDs, from left to right, indicate thestates of CC2 and CC3 respectively.

Note: The two rightmost LEDs in the top row are notassociated with any CC channels and can be ignored.


Bottom row:
The LED indicates the CC1 state.

37

Chapter 4: DX Series

DX-VBRDX-VBR contains one vibration sensor and a pair of terminals forconnecting a contact closure (CC) sensor.

Vibration sensor:
The built-in vibration sensor detects the vibration of any object where
DX-VBR is affixed. The sensor determines the vibration by measuring
the accelerations in three mutually perpendicular directions, and thengenerating a numeric value representing the magnitude of theacceleration due to gravity of the Earth. Below is the calculation formula.
a = (ax2 + ay2 + az2)1/2



The acceleration has two measurement units: g and milli-g.


1 g = 9.80665 m/s2

1 milli-g = 0.001 g"a" in the above formula denotes the magnitude of the acceleration.


0 <= a <= 3.464 g

Note: <= means "less than" or "equal to."38ax, ay and az represent the accelerations in three mutuallyperpendicular directions x, y and z.
x, y and z form a Cartesian coordinate system.

-2 g <= ax <= 2 g



-2 g <= ay <= 2 g



-2 g <= az <= 2 g

Chapter 4: DX Series

DX-VBR Terminals, dip switches and LED:

Numbers

Components
One CC channel comprising a pair of terminals.
See Connecting Detectors/Actuators to DX (on page 40)
for how to connect a CC sensor.
 Dip switch 1 configures the Normal state of the CC
channel.
 Dip switch 2 does not control any CC channel and canbe ignored.
See Adjusting Dip Switches (on page 45).
CC status LED. See Contact Closure Sensor LEDs (onpage 16) for details.
High-speed flashing of CC1 LED indicates that the DX
firmware upgrade is in progress. See Sensor Firmware
Update (on page 50).

39

Chapter 4: DX Series

Making Connections
Pre-installed DX Brackets
To allow users to hang or affix a DX onto an object or position, twobrackets have been installed on the rear side of a DX sensor package
when shipped out of the factory.
Below is the diagram of a DX sensor package with two brackets
installed.

Connecting Detectors/Actuators to DX
A DX sensor package comprises two parts: a sensor box and theterminal module(s). A terminal module is removable.
Note: The following diagrams illustrate a terminal module comprising two
termination points only. Your DX terminal module may be larger if it has
more terminals.
To make connections when the terminal module is attached:
1. Strip the insulation around 12 mm from the end of each wire of adetector or actuator.


Wire size range: AWG 28 to 20 or 0.09 to 0.5 mm2

2. Fully insert each wire into each termination point of a CC, DC or
PDC channel on the DX sensor package.

40

Chapter 4: DX Series

Important: For a PDC channel, you must check the electrical polarity
markings (+ and -) on the DX label to make sure each wire isinserted into the correct termination point with the correct polarity.

3. Use a screwdriver with a 2.5 mm wide shaft to tighten the screws
above each termination point to secure the wires, using a torque of0.196 N·m (2 kgf·cm).

To make connections after the terminal module is detached:
1. Loosen the screws at two sides of the terminal module.

41

Chapter 4: DX Series

Note: The two screws are not removable so just loosen them.

2. Separate the terminal module from the sensor box.

3. After connecting detectors/switches to the terminal module, plug theterminal module back into the sensor box, and then tighten thescrews at two sides of the terminal module.

42

Chapter 4: DX Series

Cascading DX Sensor Packages
To increase the number of connected DX sensor packages, you cancascade DX using standard network patch cables. A maximum of 12 DX
packages can be daisy chained.

Numbers Components
RJ-45 ports, each of which is located oneither end of a DX sensor package.
RJ-12 port, which is reserved for future use
and is hidden now.
To cascade DX packages:
1. Connect a standard network patch cable (CAT5e or higher) to either
RJ-45 port of a DX sensor package.
Note: For a Raritan EMX or PDU with an RJ-12 SENSOR port, use
the RJ-12 to RJ-45 adapter cable instead which is shipped with DX.
2. If you want to cascade DX packages, get an additional standard
network patch cable (CAT5e or higher) and then:a. Plug one end of the cable into the remaining RJ-45 port on theprior DX package.
b. Plug the other end into either RJ-45 port on an additional DX
package.
Repeat the same steps to cascade more DX packages.
Exception: You CANNOT cascade DX-PD2C5 sensor packages.
See DX-PD2C5 (on page 30) for details.

43Chapter 4: DX Series

3. Verify that the total number of sensors and actuators connected tothese DX packages do not exceed the maximum number of sensors
and actuators supported by your Raritan PDU or EMX. The
supported maximum number varies based on the firmware version.
See Supported Maximum Number of Sensors and Actuators (onpage 47).
Connecting a DPX2 Sensor Package to DX
You can connect only one DPX2 sensor package to the "end" of a DX
sensor chain. Simply plug the RJ-12 connector of the DPX2 sensor cable
into the RJ-45 port of the final DX in the chain.
The maximum number of DX sensor packages in the chain must be lessthan 12 when a DPX2 sensor package is involved.
When connecting a DPX2 sensor package containing three
DPX2 sensors:
A maximum of nine (9) DX sensor packages can be cascaded
because 12-3=9.

When connecting a DPX2 sensor package containing two DPX2
sensors:
A maximum of ten (10) DX sensor packages can be cascaded
because 12-2=10.

44

Chapter 4: DX Series

When connecting a DPX2 sensor package containing one DPX2
sensor:A maximum of eleven (11) DX sensor packages can be cascaded
because 12-1=11.

Adjusting Dip Switches
There are two Normal settings for each CC channel on DX packages.


N.O (Normally Open): The open status of the connected
detector/switch is considered normal.



N.C (Normally Closed): The closed status of the connected
detector/switch is considered normal.

Each CC channel and the DX-PIR occupancy sensor's Normal setting is
configured by turning on or off its corresponding dip switch.
To adjust dip switches for CC channels:


N.O: Turn ON the dip switch by pressing it down.



N.C: Turn OFF the dip switch by pressing (or keeping) it up.
To adjust the dip switch for DX-PIR occupancy sensor:

Turn it OFF (up) when human absence is considered the normal
state.



Turn it ON (down) when human presence is considered the normal
state.

45

Appendix A Supported Maximum Sensor
Distance
Every Raritan sensor package supports the total cabling length up to 98
feet (30 m), including DPX, DPX2 and DX series, except for DX-PD2C5.
You can use a Raritan sensor hub to increase the total distance of a
DPX sensor. Each sensor port on a Raritan PDU or EMX supports a
maximum of one Raritan sensor hub so DO NOT cascade sensor hubs
for distance increase.
For detailed information on DPX sensor's supported maximum distance,
see the online help for your Raritan PDU or EMX.
Note: Online help is accessible from your Raritan PDU or EMX
application, or the Support page (http://www.raritan.com/support/) on
Raritan's website.
Exception - maximum distance for DX-PD2C5:
DX-PD2C5 supports a maximum cabling length of 29 feet (9 m) insteadof 98 feet (30 m). The maximum distance must follow the following
diagram.
PX3

16 feet(5 m)

1 DX-PD2C5

13 feet(4 m)

EMKA
doorhandles

46Appendix B Supported Maximum Number of
Sensors and Actuators
The maximum number of sensors and actuators that can be managed
per sensor port of a Raritan PDU or EMX varies according to the PDU's
or EMX' firmware version.
However, DPX sensor packages are always limited to a maximum of 16
DPX sensors per port due to hardware restrictions regardless of thefirmware version.
Below lists different Raritan products and firmware versions that support
a different number of managed sensors/actuators.
Note: For the latest information on the supported maximum number of
sensors and actuators, see your Raritan PDU's or EMX' online help.
Online help is accessible from your Raritan PDU or EMX application, or
the Support page (http://www.raritan.com/support/) on Raritan's
website.
In the following list, x represents a number.
Manage up to 16 sensors/actuators per port:

PX2 series: 2.5.x and earlier

PX3 series: 2.5.x



PX3TS series: 2.6.xEMX series: 2.5.x and earlier

BCM series: 2.5.x and earlier

PXE series: 2.5.x and earlierPX, DPC and DPX (PDU) series: 1.5.x and earlier

Note: PX, DPX and DPC PDUs support DPX sensor packages only.
Manage up to 32 sensors/actuators per port:

PX2 series: 3.0.0 and later

PX3 series: 3.0.0 and later

PX3TS series: 3.0.0 and later

EMX series: 3.1.0 and later

BCM series: 3.0.0 and later

PXE series: 3.0.0 and later
Calculation examples for 16 and 32 sensors/actuators:

47

Appendix B: Supported Maximum Number of Sensors and Actuators

A sensor package may contain more than one sensor/actuator. A sensor
or actuator is a function. For example, a DX-D2C6 contains 8 functions
because it has 6 sensors and 2 actuators.
When a Raritan PDU or EMX supports 16 managed sensors or
actuators:
2*8 = 16
Therefore, you can manage two DX-D2C6 packages per port.

When a Raritan PDU or EMX supports 32 managed sensors or
actuators:
4*8 = 32
Therefore, you can manage four DX-D2C6 packages per port.
Tip for DX sensor connection:

If intended, you can connect a maximum of 12 DX sensor packages
without exceeding the maximum number of sensors/actuators supported
by your EMX or PDU. All you need to do is to manage "partial"
sensors/actuators of each DX and unmanage the rest.
For example, if your EMX or PDU can manage 32 sensors/actuators perport, you can connect 12 DX-D2C6 sensor packages to a sensor port bymanaging only two sensors or actuators of each DX-D2C6.
12*2 = 24
All 24 sensors/actuators can be managed because it is less than 32.
For information on how to manage and unmanage any sensors/actuators,
see online help or User Guide of your PDU or EMX.

48

Appendix C Sensor Measurement Accuracy
This section shows the factory specifications of Raritan DPX and DPX2
environmental sensor packages. Calibration is not required for thesesensor packages.

In This Chapter
DPX Sensor Accuracy .............................................................................49
DPX2 Sensor Accuracy ...........................................................................49

DPX Sensor AccuracyTemperature: +/-2 degrees Celsius (+/-3.6 degrees Fahrenheit)Humidity: +/-5% (when humidity < 60%) or +/-8% (when humidity >60%)Differential air pressure: +/-1.5%

Air flow: +/-6.5%

DPX2 Sensor AccuracyTemperature: +/-1.0 degree Celsius (+/-1.8 degrees Fahrenheit)Humidity: +/-2.0% (when humidity = 10% to 90%)49

Appendix D Sensor Firmware Update
DPX2 and DX sensor packages automatically upgrade or downgrade
their firmware after being connected to a Raritan EMX or PDU that
supports these sensor packages. This way it ensures that these sensor
packages work properly with the connected Raritan PDU or EMX.
Automatic upgrade or downgrade is determined by comparing DPX2 or
DX sensor firmware against the environmental sensor firmware stored
inside the Raritan PDU or EMX. A downgrade can be performed only
when the PDU- or EMX-provided sensor firmware is either greater thanor equal to the minimum firmware version requested by the connected
sensor packages.
Note that a DPX sensor package does NOT support the automatic
firmware update feature, and you do NOT need to manually update it.
Tip: Information regarding current sensor firmware version and thesensor's firmware update time is available in the Device Information
dialog. Choose Maintenance > Device Information > Peripheral Devices
in the web interface of your Raritan PDU or EMX.

50

Appendix E Sensor Naming Conventions
The model name of a Raritan environmental sensor package consists ofsensor family, sensor functions and the total number of the specified
function.
[Family] - [Function_1] [Number_1]- OR [Family] - [Function_1] [Number_1] [Function_2][Number_2][Family] is DPX, DPX2 or DX.
[Number_1] and [Number_2] are integer numbers.
[Function_1] and [Function_2] are abbreviations representing diverse
functions.
Note: Some sensor packages may indicate the main function only rather
than indicating all functions in its name. For example, the name of
DX-VBR indicates that it contains the vibration sensor while it does notindicate the contact closure sensor that it also contains.
Abbreviations
AF
CC or C
D
DP
H
PD
PIR
T
VBR
WSC or WC
WSF

Functions
Air flow sensors
Contact closure sensor terminals
Dry contact signal terminals for actuators
Differential air pressure sensors
Humidity sensors"Powered" dry contact signal terminals
Passive infrared occupancy sensor
Temperature sensors
Vibration sensors
Cable water sensors
Floor-mounted water sensors

51

IndexA

F

Adjusting Dip Switches • 30, 33, 37, 38, 41, 47
Air Flow Sensors • 5, 6
Available DPX Sensor Packages • 2, 5
Available DPX2 Sensor Packages • 2, 21
Available DX Sensor Packages • 2, 28

Floor Water Sensors • 17
Floor-mounted Water Sensor • 5, 17

C
Cable Water Sensor • 5, 18
Cascading Air Pressure Sensors • 7, 8
Cascading DX Sensor Packages • 45
Configuring a Contact Closure Sensor • 14, 16
Connecting a DPX2 Sensor Cable • 24
Connecting a DPX2 Sensor Package to DX •46
Connecting Detectors/Actuators to DX • 30, 33,
37, 38, 41, 42
Connecting Detectors/Switches to New
DPX-CC2-TR • 10, 11
Connecting Detectors/Switches to Old
DPX-CC2-TR • 10, 11, 13
Connecting Tubes • 8Connection and Disconnection • 23
Contact Closure Sensor LEDs • 16, 31, 34, 37,
39, 41
Contact Closure Sensors • 5, 9D
Differential Air Pressure Sensors • 5, 7
Disconnecting a DPX2 Sensor • 26
DPX Sensor Accuracy • 51
DPX Series • 4
DPX Temperature and Humidity Sensors • 5,
6
DPX2 LED States • 22
DPX2 Sensor Accuracy • 51
DPX2 Series • 20
DPX2 Temperature and Humidity Sensors •21, 22
DX Series • 16, 27
DX-D2C6 • 28, 29
DX-D4C3 • 28, 37
DX-PD2C5 • 28, 32, 45
DX-PIR • 28, 35
DX-VBR • 28, 40

I
Introduction to Environmental Sensor
Packages • 1

MMaking Connections • 42

O
Old and New Contact Closure Sensors • 11

P
Pre-installed DX Brackets • 42

S
Sensor Comparison • 2
Sensor Firmware Update • 22, 31, 34, 37, 39,
41, 52
Sensor Measurement Accuracy • 51
Sensor Naming Conventions • 5, 21, 28, 53
Sensor Overview • 2
Supported Maximum Number of Sensors andActuators • 1, 2, 8, 46, 49
Supported Maximum Sensor Distance • 2, 32,
48

53

U.S./Canada/Latin America
Monday - Friday
8 a.m. - 6 p.m. ET
Phone: 800-724-8090 or 732-764-8886
For CommandCenter NOC: Press 6, then Press 1
For CommandCenter Secure Gateway: Press 6, then Press 2
Fax: 732-764-8887
Email for CommandCenter NOC: tech-ccnoc@raritan.com
Email for all other products: tech@raritan.com

China

EuropeEuropeMonday - Friday
8:30 a.m. - 5 p.m. GMT+1 CET
Phone: +31-10-2844040
Email: tech.europe@raritan.com

United Kingdom
Monday - Friday
8:30 a.m. to 5 p.m. GMT
Phone +44(0)20-7090-1390

Beijing

FranceMonday - Friday
9 a.m. - 6 p.m. local time
Phone: +86-10-88091890

Monday - Friday
8:30 a.m. - 5 p.m. GMT+1 CET
Phone: +33-1-47-56-20-39

Shanghai

Germany

Monday - Friday
9 a.m. - 6 p.m. local time
Phone: +86-21-5425-2499

Monday - Friday
8:30 a.m. - 5:30 p.m. GMT+1 CET
Phone: +49-20-17-47-98-0
Email: rg-support@raritan.com

GuangZhou
Monday - Friday
9 a.m. - 6 p.m. local time
Phone: +86-20-8755-5561

India
Monday - Friday
9 a.m. - 6 p.m. local time
Phone: +91-124-410-7881

Japan
Monday - Friday
9:30 a.m. - 5:30 p.m. local time
Phone: +81-3-5795-3170
Email: support.japan@raritan.com

Melbourne, Australia
Monday - Friday
9:00 a.m. - 6 p.m. local time
Phone: +61-3-9866-6887

TaiwanMonday - Friday
9 a.m. - 6 p.m. GMT -5 Standard -4 Daylight
Phone: +886-2-8919-1333
Email: support.apac@raritan.com