Application Note

Time Domain Measurements
Using Vector Network Analyzers
MS4640 Series
VectorStar™ VNA

Introduction
Vector Network Analyzers (VNAs) are very powerful and flexible measuring instruments. Their basic capability is
to measure the S-parameters of an RF or microwave device and display the result in the frequency domain. This
provides valuable data for the design engineer to develop a design and for the production engineer to substantiate
the performance of the device or system. In addition to displaying performance in the frequency domain, Anritsu
VNAs offer a Time Domain option to enhance the analysis capability by presenting data in the time (or distance)
domain–valuable information for both the design and production engineer. Figure 1 provides the time andfrequency response of a microstrip circuit as measured by the MS4640 VNA Series.
Note that port 2 of the DUT is left open and consequently the frequency plot of the device provides little information
about the performance of the device. The benefit of the Time Domain response is in providing information of thedevice performance at specific locations. As a result, the VNA provides the overall S-parameter critical performance
information for the device both as it appears when installed in a system and also at a specific location within the device.

Time Domain Fundamentals
The Fourier transform has been used in many technologies and in microwave measurements. It provides a method fortransforming frequency domain data, which is the VNA’s basic capability, into the time domain. Microprocessors arefundamental to modern VNAs by providing an efficient method of computing error correction terms. In addition,
micro-processing also permits convenient implementation of the Fast Fourier Transform (FFT) for additional
advanced analysis.
Specifically, a variation of the chirp z inverse FFT is used to transform from the frequency domain to the time
domain. This permits the user to “zoom in” on a specific time (distance) range of interest for the data display. Thisapplication note addresses the properties of the transform process and shows how the various processing options
are used to obtain optimum results for a given application. It will not address the algorithmic details of the FFT.
Microstrip
Coax

Coax
Port 2

Port 1Figure 1. Time and Frequency Response of a Microstrip Circuit.

2

A convenient starting point is to consider the transform of frequency domain data obtained from the measurementof a short at the end of an airline as shown in Figures 2a and 2b. It is apparent that the time domain response isinformative; but it is far from the ideal time domain response that is superimposed. This figure illustrates two of thefundamental properties (limitations) associated with the FFT: resolution and alias free range. Resolution is limitedby the bandwidth of the measurement whereas aliasing is determined by the frequency step size.

Figure 2a.

Alias

Figure 2b.

3

The principal property of interest in time domain processing for most microwave applications is resolution, or theability to resolve one location from another. In practice, the basic limitation is inversely related to data collection
bandwidth in the frequency domain. A rule of thumb defines resolution on the order of 150 mm (for air dielectric)
divided by the Frequency Span (GHz) . For example, a 20 GHz frequency span will provide a resolution of about 8 mm
while a span of 70 GHz will provide a resolution of about 2 mm. So the broader the frequency span, the more
information presented during the time domain analysis. Figure 3 illustrates the importance of a wide frequency sweep
if the application calls for identifying closely spaced reflections. Note that even though the DUT may not be required
to perform up to 70 GHz, as long as the DUT doesn’t completely fall apart out of band, the design engineer can still
benefit from the 70 GHz span due to the increased resolution provided by the wide frequency range. A limitation
to the frequency span rule occurs when measuring devices with limited bandwidth. Ideally, the device will pass the
DC component, and as a result, the performance at higher frequencies will result in a gradual roll-off. If the DUT
does not pass the DC component, such as band pass or high pass filter, then resolution will degrade.
Resolution is also influenced by the processing method and window selection options that are described below.
Another important factor is the relative amplitude of the signals being processed. A large signal can swamp asmaller signal–where the greater the time (distance) separation between the two discontinuities, the better the
points of interest can be resolved.
TWO MISMATCHES SEPARATED
By 2 mm (AIR)SPAN RESOLUTION
1) 40 GHz 3.75 mm2) 50 GHz 3.0 mm
3) 60 GHz 2.5 mm

Figure 3. Wide bandwidth provides improved discontinuity resolution.

4

Figure 4 represents a time domain plot of a device with two discontinuities separated by 4.8 mm. The discontinuities
are located at marker 3 and the display is taken from a VNA whose maximum bandwidth is 26.5 GHz. Low passprocessing, the highest resolution mode possible, is used. Note that the 26.5 GHz frequency span is unable toresolve the two discontinuities.
Figure 5 provides a low pass plot of the same device but with a VNA capable of covering a 65 GHz frequency span.
Markers 2 and 4 on the 65 GHz display are at the same location as in the 26.5 GHz plot. Note the additional amountof information available from the wider bandwidth. Also note the difference in return loss value when compared to anarrow band sweep. This points out that for maximum resolution and accuracy, the widest frequency span available
is best for optimum time domain analysis, especially when resolving two discontinuities in close proximity.

Figure 4. Low pass time domain plot of two discontinuities 4.8mm apart located at marker 3.
VNA has an upper bandwidth limitation of 26.5 GHz.

Figure 5. Same device measured with 65 GHz bandwidth. Second discontinuity
now visible at marker 6. Note difference in return loss values.

5A property of the transform process is the Alias Free Range. The transform is a circular function and repeats itself
outside of its inherent range that is t = 1/(Frequency Step Size) The frequency step size is proportional to thefrequency span and inversely proportional to the number of data points (N).
Inherent time range: t = (N-1)/(Frequency Span)
For example, with a 40 GHz frequency span and 1,001 data points, the Alias Free Range is 1000/40 GHz =25 nanoseconds. This is large for most applications, and the range is usually greatly reduced for display. However,if the user wanted to locate a fault in a 100 meter cable, the range is inadequate. For such applications the step
size must be reduced either by decreasing the frequency span or increasing the number of points.
A wide alias free range is required when analyzing long transmission cables such as optical cables or microwave
transmission lines.

Processing Alternatives
The Time Domain option offers a number of processing alternatives to the user. It is important that the user beaware of the feature set available as selection can have a significant effect on the end result.
These tools include:• Processing Method – Low pass or Band pass
• Response – Step or Impulse for Low pass mode and Impulse or Phasor Impulse for Band pass mode
• Window Shape – Rectangular, Nominal, Low Side lobe, Min Side lobe
• Gating – Frequency with Time Gate

Low pass Processing
This processing method is useful in determining the characteristics of a discontinuity or component. It has twice
the spatial resolution of the band pass mode. The DC value must be approximated by extrapolation, and harmonic
calibration is required. The frequency plan for the Low pass mode is shown in Figure 6.
• The time domain low-pass processing simulates the traditional TDR measurement and supports both thestep and impulse response.
• Harmonic frequency VNA measurements are required starting at the lowest frequency (FL).
• The DC term is determined by data extrapolation from the low end data points.
• It is assumed that the frequency response is hermitian (in that the negative frequency response is the
conjugate of the positive frequency response). Thus the Time Domain response is real and not complex.
• The shape of the real part of the step and impulse response in the Time Domain will show the nature of thecomplex discontinuity.
The lower the start frequency of the VNA, the more accurate the DC extrapolation calculation. With a start frequencyof 70 kHz, the MS4640 series offers the lowest available DC extrapolation point thereby reducing the DC data point
error. When additional information is available regarding the test environment, the DC term can be entered manually.

• VNA data set = n*FL
• Where: n = 0,1,2,….N,• N = FH /FL,
• # of data points = N+1• frequency step ∆F = FL
• n = 0 (DC) term calculated
by extrapolation
–freq
–FH

freq
FH

DC
–FL FL
Total Bandwifdth = 2FH

Figure 6. Total Frequency Spectrums for Low Pass Processing.

6

As mentioned above, Low Pass offers two presentations: Step or Impulse response. The Step response permits
the direct display of impedance versus time or distance, similar to that obtained with a TDR. The Step response
display is the integral of the Impulse response. The user can also obtain insight into the nature of a discontinuity:
inductive, capacitive or resistive. Figure 7 shows ideal responses associated with various types of discontinuities.
The default display graph type is real and shows impedance information (a scale of 20 mU per division is about 2 ohms
per division). The user can choose Log Mag or a linear display format. Low pass processing also offers the highest
resolution for a given bandwidth and provides twice the resolution of Band pass mode. If the instrumentation, thetransmission line, and the DUT permit Low pass processing, it should be used.

Figure 7. Ideal responses associated with various types of discontinuities.

Figure 8 shows a practical application of Low pass Time Domain processing. An SMA line standard wasconstructed using solid Teflon® and precision connector interfaces. The response clearly shows the quality of the
SMA “standard” that was used to evaluate SMA calibration performance. When considering time domain display,
the most common method of display is in linear mode. So the display of Figure 8 is in terms of Reflection
Coefficient, ρ (VSWR can also be used). The primary reason for displaying the response in linear units is theability to display the impedance of the DUT. The standard Air-line has a reflection coefficient of –2.679 mU or–51.44 dB. The VSWR (ρ) = 1.0054 and the impedance = 49.73 ohms (R = Zo/VSWR).

Figure 8. Practical application of Low pass Time Domain processing.

Summary of Low-pass Processing• Displays similar to traditional TDR• DC value determined by extrapolation• Frequency plan is harmonically related to the start frequency• Spatial resolution is twice that of Band-pass processing• “Reflection Step Response” useful for location and identification of discontinuities for devices that exhibit
Low-pass response
Teflon is a registered trademark of the DuPont Company

7

Band Pass Processing
Many situations, such as waveguide or band limited DUTs, preclude using a broadband harmonically related frequency
plan. For these cases, Band pass processing should be used. Since the Band pass mode does not include a DC
value, only the impulse excitation is available. The location of the time domain impulse can be used to locate
discontinuities and faults on transmission lines. The frequency plan for Band pass processing is shown in Figure 9.
Total Bandwidth = FH –FL• Frequency step:• F(n) = FL+n (FH –FL )/N,
• Where: n = 0,1,2,3..N• Number of data points = (N+1)• Freq. step = (FH –FL )/N• Bandwidth = (FH –FL )
freq

DC

freq
FH

FL

Figure 9. Total Frequency Spectrum for Band Pass Processing.

Summary of Band-Pass Processing• Band Pass mode does not include the DC term, thus only Impulse response is available• Any linear frequency calibration in span FL to FH• Band limited data
• Scalar information only (Location), does not give impedance
Figure 10 shows the Band pass response of the SMA “standard” that was used in Figure 7. This type of processing iscommon in fault location that can be used to test transmission lines.
Figure 11 shows an example of this application: measuring a 2 m cable with imperfections.

Figure 10. The Band pass response of a SMA ‘Standard.

8

Figure 11. Measuring a 2 m cable with imperfections.

Summary for Band-pass processing• Only Band-pass Impulse mode is useful• The frequency plan is an arbitrary linearly related frequency set starting at the low end frequency
FL and ending at the high end frequency FH• This mode is mainly used for fault location• It will show reflection and transmission magnitude

Phasor Impulse
This technique, exclusive to Anritsu VNAs, enables the user to extract impedance information for a specificdiscontinuity from a band-pass display. The discontinuity of interest is positioned in the center of the screen andshould fill about 50% of the time range being displayed. When the Phasor Impulse processing is applied, theresulting display, usually Real, Imaginary, or Linear Polar, provides the desired impedance information.Figure 12
shows the shunt capacitance associated with a waveguide iris.

Figure 12. Normal Band-pass display and the Phasor Impulse display of waveguide.
9

Figure 13 is a summary of the processing types. As noted, Low-pass processing should be used whenever possible
as it offers the highest performance and most versatile set of displays.

Figure13. Summary of processing types.

10Windows
Windows are used to “condition” data prior to transformation. They are used to get around the Fourier transforms basic,
but impractical requirement to use frequency terms stretching from –∞ to +∞. They are used to mitigate edge effects
associated with a finite data set. The application of what is called a “window” function to the frequency domain data
will accomplish this. The window functions are curves derived from a mathematical function (many are available–
see reference 4) that tapers off from unity gain at the center of the frequency domain data to a low value at theends. Figures 14a and 14b show the effect of applying the widely used two-term Hamming window to the dataarray obtained by measuring the short circuit shown earlier. The resulting time domain displays are shown in 15a
and 15b. This is not without a price, since the window has the effect of widening the main lobe, thus decreasing
the effective resolution. For this reason, it is good to have a range of different window types available so thatmultiple views between resolution and side lobe effects, appropriate for the application, can be used. The Anritsu
VNAs offer the window choices described in Table 1. Figure 16 provides a display of window effects on main andside lobes. Resolution has been normalized to the Default Nominal window that is adequate for most applications.

Figure 14a. Frequency domain data before application of Hamming Window.

Figure 14b. Frequency domain data after application of Hamming Window.

11

Figure 15a. Time domain data before application of Hamming Window.

Figure 15b. Time domain data after application of Hamming Window.

Window Type

Main Lobe Width Lowpass

Main Lobe Width Bandpass

Sidelobe Level (dB)

Rectangular

0.5/BW

1/BW-13

Nominal

1/BW

2/BW

-42

Low sidelobe

1.5/BW

3/BW-70

Min sidelobe

2/BW

4/BW

-90

Table 1.

12

In addition to the standard selection of window shapes, advance parameters unique to the MS4640A VNA are
now available. The advanced control is available using the new selections, Kaiser-Bessel and Dolph-Chebyshev.
Figure 16 provides an example of control possibilities using these choices. These two new window types allow fora finer selection of the trade-off between side lobe level and resolution. For the Kaiser-Bessel window, a larger
Beta value leads to lower side lobes, and a wider main lobe width. For the Dolph-Chebysev window, the side lobelevel is parameterized explicitly (in absolute dB) and a larger value leads to lower side lobes. The window for two
parameter values for each of these windows is shown in Figure 17 along with the rectangular window for comparison.
Low-pass Time Domain Window Effects
0
–10
–20

dB

–30

rectangularnominalside lobe minside lobe

–40
–50
–60
–70
–80
50

60

70

80

90

100

110

120

130

140

150

Time (ps)Figure 16. Window effects on main and side lobes.

Low-pass Time Domain Window Effects
0
–10
–20

dB

–30

rectangular
Kaiser beta=6
Kaiser beta=10
dolphSL=40
dolphSL=80

–40
–50
–60
–70
–80

50

60

70

80

90

100

110

120

130

140

Time (ps)Figure 17. Advanced Time Domain controls on main and side lobe display.

13

150

The approximate relationship between these parameters and the main lobe width (null-to-null) is suggested in
Figure 18. Here, everything is scaled relative to a rectangular window (a nominal window is at 2, a low side-lobe
window is at 3, and a minimum side-lobe window is at 4 on this scale) and the y-axis is normalized relative to thelobe width of a rectangular window.
Main Lobe Width vs. Advanced Windo Parameters
7
Kaiser-Bessel

6
Relative main lobe width

Dolph-Chebyshev
5
4
3
2
1
0
0

10

20

30

40

50

60

70

80

90

100

110

120

Beta (unit less) or side lobe level (dB)Figure 18. Relative relationship of main lobe between
Kaiser-Bessel and Dolph-Chebyshev parameters.

Gating – Frequency Gated by TimeThe Gate is essentially a filter in the time domain. It is set by the operator while observing the time domain response
to select a given discontinuity or in some cases to select those responses that can be attributed to a given DUT,
such as a cable that has principal reflections at both ends. It can also be used to Gate out a specific discontinuity,
such as a test fixture, thus observing the performance of a microwave circuit with an imperfection removed. Thispermits observing the selected range without the influence of unwanted elements, such as connectors or in thecase of transmission measurements: multi-path signals. The Gating process also involves selecting a gate shape.
This is similar to, but not identical to, the Window function. In looking at previous time domain displays, it isapparent that there are ripples associated with a specific reflection that extend over the time range. All of these arenecessary to obtain a good representation of that reflection in the frequency domain. The gating process caneliminate some of this necessary data that can introduce errors or ripple in the frequency domain response. As in
Windowing, the solution is to provide a selection of Gate functions so the user can obtain the best tradeoff ofresolution versus ripple.
Figure 19 shows the application of a Gate and the Frequency Gated by Time display of the initial discontinuity of
the Beatty standard used in Figure 13. This figure also illustrates the processing sequence that can be observed
when a four trace display is available. Each trace is set to S11 and each is set up with a different process, the final
in the lower right showing the frequency gated by time response of an impedance step that is reasonably flat withfrequency as expected.

14

Figure 19. The application of a Gate and the Frequency Gated by Time display of
the initial discontinuity of the Beatty Standard.

Time Domain Applications
Time Domain is most useful when analyzing broadband low loss devices such as transmission lines, connectors,
test fixtures and similar components. Narrowband devices such as filters, isolators or matching networks are notgood candidates for Time Domain measurements.
A few “Helpful Hints”:• A harmonic frequency set is most desirable as this permits low pass processing. In any application, use thewidest frequency range possible and start as low as possible for best DC extrapolation.
• When interpreting the return loss value of a Time Domain response, it must be noted that the response is theaverage of the return loss over the entire frequency range.
• When using frequency gated by time, it is important to center the gate around the discontinuity or circuit
being analyzed. In general, the gate should be as wide as possible, but not too close to the responses thatthe user wants to eliminate.
• The selection of Window and Gate shapes can have a significant effect on the result. A wide selection range
provides the opportunity to optimize results. In these cases, if correlation with other measurements becomes
an issue, it is important that the same Window and Gate shapes are used.

15

Utilizing Airlines
Time Domain work is greatly enhanced when the reflection to be studied is well separated from other reflections.
This is readily accomplished if a high quality transmission line can be included in the system to be measured.
In a coaxial system this would be an airline with connectors that interface to other components with minimum
reflections. This performance can be realized by airlines with a beaded interface at one end, and a floating centerinterface at the other end. The bead insures an accurate pin depth at the precision interface, and also supports thecenter conductor so the airline can be easily installed.
The following examples show a number of practical applications for Time Domain processing:

Designing a Test Fixture
In test fixture design, it is often desirable to have an input coaxial connector and a launch to the microstrip medium
being used for circuit design. This launch should be designed for minimum SWR so that it will be transparent when
the device design is being evaluated. This can be accomplished by starting with a short length of microstrip line
(several centimeters is a good choice) and connecting it to the initial launcher design, leaving the other end open.
Figure 20 shows the forward reflection of a coaxial to Duroid microstrip design. The reflection from the launch isclearly distinguished from the open end. The designer can make modifications to the launch design and easily see
their effect until a satisfactory launch interface is obtained.

Figure 20. Coaxial to Duroid microstrip design.

16

Measurement of VNA Directivity and Port Match
The accuracy of a VNA reflection measurement is primarily related to the effective directivity and port match after
calibration. These parameters are usually specified in data sheets, but are dependent upon calibration standards
and user technique. In some cases it would be important to know the value of these parameters. This information
can be readily obtained if an Assurance Airline is available. It is installed at the calibrated test port and terminatedwith a short circuit. Figure 21 shows by diagram what is happening; the port match, directivity, and the shortreflection are clearly separated in time and can be distinguished and evaluated quantitatively. The results show theresult of an SOLT calibration using a broadband termination. Keep in mind the values displayed are a weighted
average across the frequency sweep range.

Figure 21. Port match, directivity and short reflection clearly separated in time.

17

Rise Time Measurements using Time Domain Transmission
The advent of data transfer rates in the tens of GHz region has created the need to measure the rise time andpropagation delay of wide band devices such as coaxial cables, switches, and amplifiers used in optoelectronicsystems. The use of sampling oscilloscopes in conjunction with fast rise pulse generators is a common techniquefor measuring rise time and delay. However, as VNA bandwidths have surpassed those of the scopes, the VNA
provides an alternative means of performing the rise time and propagation delay measurements of these highspeed devices.
These measurements require the use of Time Domain Transmission that is an S21 Step response. The VNA
performs these measurements with rise times as fast as 9 picoseconds (110 GHz system). The processing options(low-pass, band-pass, and the gating feature discussed earlier) apply to S21 Time Domain response as well.
In addition, the error correction techniques available on the VNA allow more accurate measurements of rise time,
which may be obscured by the adapters and cables which are usually uncorrected in the oscilloscope method.
The Rise Time measurement of a 40 Gb/s driver amplifier using the 110 GHz VNA is discussed in this section.
A measurement of the rise time of the VNA transmission response is required in order to make the amplifiermeasurement. In this case, a 12-term calibration is performed, and with the thru line in place, the S21 time step
response is measured as shown in Figure 22. The rise time, the time between the 10% and the 90% magnitude
points, is measured to be approximately 9 ps.

Figure 22. Corrected Step response of the 110 GHz VNA.

18

The 40 Gb/s driver amplifier is then measured (Figure 23). The S21 Time Domain response in this case is thecomposite measurement of the VNA rise time and the Amplifier rise time. The 180° phase shift in the amplifier S21
response is displayed as an inverted step in Time Domain mode. The composite rise time measured is 12 ps.

Figure 23. Inverted rise time and propagation display response of the amplifier.

Rise time of the amplifier can then be calculated by using the root sum squares theorem:
TR2 (DUT) = TR2 (total) - TR2 (input step)Or, TR (amplifier) = √ [(12)2 -(9)2]= 8 psSimilarly, the propagation delay through the amplifier can be measured from the 50% point of the step response.
Time Delay Td (amplifier) = 30 ps

References
1. M. Hines and H. Stineheifer, “Time Domain Oscillograph Microwave Network Analysis Using Frequency
Domain Data IEEE Transactions on Microwave Theory and Techniques”, V MTT-22, No. 3, Mar. 1974
2. L.R. Rabiner, “Chirp z-Transform Algorithm Program”, Programs for Digital Signal Processing, Ed. by the
Digital Sign Processing Committee, IEEE Acoustics, Search and Sign Processing Society, New York, IEEE
Press, 1979
3. L.R. Rabiner and B. Gold, “Theory and Application of Digital Signal Processing”, Prentice-Hall, Englewood
Cliffs, NJ, 1975
4. Fredric J. Harris, “On the Use of Windows for Harmonic Analysis with the Discrete Fourier Transform”, inProceedings of the IEEE Vol. 66, No. 1, pp 51-83, Jan. 1978
5. N. Marcuvitz, Waveguide Handbook, New York, McGraw-Hill, 1951
6. VectorStar MS4640A Microwave Vector Network Analyzer Measurement Guide

19• United States
Anritsu Company• Italy
Anritsu S.r.l.

• Singapore
Anritsu Pte. Ltd.

1155 East Collins Blvd., Suite 100, Richardson,
TX 75081, U.S.A.
Toll Free: 1-800-267-4878
Phone: +1-972-644-1777
Fax: +1-972-671-1877

Via Elio Vittorini 129, 00144 Roma, Italy
Phone: +39-6-509-9711
Fax: +39-6-502-2425

60 Alexandra Terrace, #02-08, The Comtech (Lobby A)
Singapore 118502
Phone: +65-6282-2400
Fax: +65-6282-2533• Canada
Anritsu Electronics Ltd.
700 Silver Seven Road, Suite 120, Kanata,
Ontario K2V 1C3, Canada
Phone: +1-613-591-2003
Fax: +1-613-591-1006• SwedenAnritsu AB
Borgarfjordsgatan 13A, 164 40 KISTA, Sweden
Phone: +46-8-534-707-00
Fax: +46-8-534-707-30• Finland
Anritsu AB• Brazil
Anritsu Eletrônica Ltda.

Teknobulevardi 3-5, FI-01530 VANTAA, Finland
Phone: +358-20-741-8100
Fax: +358-20-741-8111

Praça Amadeu Amaral, 27 - 1 Andar
01327-010 - Bela Vista - São Paulo - SP - Brazil
Phone: +55-11-3283-2511
Fax: +55-11-3288-6940• DenmarkAnritsu A/S (Service Assurance)
Anritsu AB (Test & Measurement)• Mexico
Anritsu Company, S.A. de C.V.
Av. Ejército Nacional No. 579 Piso 9, Col. Granada
11520 México, D.F., México
Phone: +52-55-1101-2370
Fax: +52-55-5254-3147• United Kingdom
Anritsu EMEA Ltd.
200 Capability Green, Luton, Bedfordshire, LU1 3LU, U.K.
Phone: +44-1582-433280
Fax: +44-1582-731303• France
Anritsu S.A.

Kay Fiskers Plads 9, 2300 Copenhagen S, Denmark
Phone: +45-7211-2200
Fax: +45-7211-2210• Russia
Anritsu EMEA Ltd.
Representation Office in Russia
Tverskaya str. 16/2, bld. 1, 7th floor.
Russia, 125009, Moscow
Phone: +7-495-363-1694
Fax: +7-495-935-8962• United Arab Emirates
Anritsu EMEA Ltd.

12 avenue du Québec, Bâtiment Iris 1- Silic 612,
91140 VILLEBON SUR YVETTE, France
Phone: +33-1-60-92-15-50
Fax: +33-1-64-46-10-65

Dubai Liaison Office
P O Box 500413 - Dubai Internet City
Al Thuraya Building, Tower 1, Suite 701, 7th Floor
Dubai, United Arab Emirates
Phone: +971-4-3670352
Fax: +971-4-3688460• Germany
Anritsu GmbH

• India
Anritsu India Private Limited

Nemetschek Haus, Konrad-Zuse-Platz 1
81829 München, Germany
Phone: +49-89-442308-0
Fax: +49-89-442308-55

2nd & 3rd Floor, #837/1, Binnamangla 1st Stage,
Indiranagar, 100ft Road, Bangalore - 560038, India
Phone: +91-80-4058-1300
Fax: +91-80-4058-1301®Anritsu All trademarks are registered trademarks of
their respective companies. Data subject to change
without notice. For the most recent specifications visit:www.anritsu.com• P.R. China (Shanghai)
Anritsu (China) Co., Ltd.
Room 2701-2705, Tower A,
New Caohejing International Business Center
No. 391 Gui Ping Road Shanghai, 200233, P.R. China
Phone: +86-21-6237-0898
Fax: +86-21-6237-0899• P.R. China (Hong Kong)
Anritsu Company Ltd.
Unit 1006-7, 10/F., Greenfield Tower, Concordia Plaza,No. 1 Science Museum Road, Tsim Sha Tsui East,
Kowloon, Hong Kong, P.R. China
Phone: +852-2301-4980
Fax: +852-2301-3545• JapanAnritsu Corporation
8-5, Tamura-cho, Atsugi-shi, Kanagawa, 243-0016 Japan
Phone: +81-46-296-1221
Fax: +81-46-296-1238• Korea
Anritsu Corporation, Ltd.
502, 5FL H-Square N B/D, 681
Sampyeong-dong, Bundang-gu, Seongnam-si,
Gyeonggi-do, 463-400 Korea
Phone: +82-31-696-7750
Fax: +82-31-696-7751• Australia
Anritsu Pty. Ltd.
Unit 21/270 Ferntree Gully Road, Notting Hill,
Victoria 3168, Australia
Phone: +61-3-9558-8177
Fax: +61-3-9558-8255• Taiwan
Anritsu Company Inc.
7F, No. 316, Sec. 1, NeiHu Rd., Taipei 114, Taiwan
Phone: +886-2-8751-1816
Fax: +886-2-8751-1817

11410-00722, Rev. A Printed in United States 2013-08©2013 Anritsu Company. All Rights Reserved