Cℓ−Reference electrode, made from Glass,
ISAB not required for calibration,
Newly developed long-life electrode.

Range：0～2000mg/ℓ

ISAB = Ion strength adjustment buffer
Low calibration

High calibration※ISAB not required for calibration

Measurement

For test water with low ionicstrength, add ISAB and measure

MemoryCleaning and Storage

Press the MEMO. key to store themeasured value.Indication ofmemorized value with the ▲▼ keys.

After measurement, wash the electrodewell with pure water or wipe the cloth.
⇒Store the electrode with the rubber cap.

Long-life chloride ion electrode with excellent stability and durability

Chloride ion meter

CL-11ZInstruments◆High efficiency & Long-life electrodeThe new type chloride ion electrode consisting of a reference electrode made fromglass and special electrode inside liquid, has excellent stability and durability.

◆The reference electrode is glass of Long-life. Hg free
The reference electrode is glass, making it durable
and resistant againt contamination.

ModelObject
Range

Resolution

Temp. compensation

Repeatability◆ISAB not required for calibration
Special electrode inner liquid and calibration liquid with increased ion strength.

◆Linearity of electrode output characteristics.

Cl-：0～2000mg/L

mV：0～±1000mV（for electrode inspection）

Temp.

：0.0～50.0℃（Recommended Use 6～45℃）－Cl mode：0.1mg/L （0.1～99.9mg/L）：1mg/L （100～2000mg/L）

mV mode
Temp.

：1ｍV
：0.1℃

Automatic by thermistor
Chloride ion：Within ±2％（For equivalent input）

Temp.

：Within ±0.5℃ （For equivalent input）

Disconnection

Self diagnosis
function

Calibration ERR

Memory function

Max. 30 dataPowerStructure

Excellent linearity from low concentration to high concentration

CL-11Z（Not compatible with CL-10Z）
Water supply, food and beverage, septic tanks, industrial wastewater, etc.

Outer dimension：S ERR：CAL ERR：Battery mark blinks

Battery shortage

DC4.5V（LR03 battery ×3）
Auto power off system(30 minutes)
Dust-proof and water-proof performance equivalent to IP67

（The detector be properly connected to the instrument.）70(W)×40(H)×173(D) mm

Electrode
Name
Model
MethodElectrodeconfiguration
Wetted partTemp. condition
Size／Weight◆Use

Selectivity

Sewage, human waste, septic tank, food, chemical factory, waterworks

New type Chloride ion electrodeCLD-11（Not compatible with CL-2206）
Chloride ion electrode
Working electrode：AgCl
Refelence electrode ：Glass
PP、AgCl、PVC、ABS、Ceramic
0～45℃
φ16.5×187.5、About 110ｇ
S２－・・・・・Coexistence is not possible
CN－、I－・・・・・・10-5
Br-、S2O32-・・・・・・10-2
NO3-、SO42-、CO32-、PO43-、F-・・・・・103

Cable length

1ｍStandard components
Main body（CL-11Z）、Chloride ion electrode（CLD-11）：CLSLN-10-50 (50mL)
10mg/L Standard solution Cl 1000mg/L Standard solution Cl －◆The three parameters measurement【Cℓ 、℃、mV】
Measure 3 parameter :chloride ion density / water temperature / electrode output (mV)

：CLSLN-1000-50 (50mL)：CLDLQ-30 (30mL)
Inner liquid for CLD-11※Not compatible with CL-10Z：50mL
Ion strength adjustment buffer (ISAB)
Pipette、Sand paper、Vinyl cover、carrying case

OptionSpare Chloride Ion Electrode （CLD-11）

MODE

Measured values

Electrode output (mV)Mode switching example

本社埼玉県久喜市桜田2丁目133番地8 〒340-0203

TEL：0480-38-9151（代） FAX：0480-38-9157

URL
KASAHARA

https://www.krkjpn.co.jp

CHEMICALINSTRUMENTS

CORP.

2-133-8 Sakurada, Kuki-City, Saitama, Japan 〒340-0203

Calibration solution 10mg/L ClCalibration solution 1000mg/L ClInner liquid for CLD-11
ISAB (=Ion strength adjustment buffer)：CLSLN-10-250 (250mL)：CLSLN-1000-250 (250mL)：CLDLQ-50 (50mL)： 250mL