1500 SERIES

N95 HEALTHCARE AND SURGICAL MASKS
NON-LATEX RESPIRATOR AND STRAPS
The non-latex straps don’t irritate or require prestretching.
Available in 5 color coded sizes.

1510 Extra Small

DURA-MESH® SHELL LASTS LONGER
The exclusive Dura-Mesh shell protects the filter media so it stays
cleaner-looking longer. The shell also resists collapsing in hot,humid environments so the respirator holds its shape longer. The
result is fewer respirators used and lower costs.
SOFT FOAM NOSE CUSHION
The molded nose bridge with soft foam nose cushion provides
an easy comfortable fit. No metal noseband to adjust and seal
is maintained without pressure points.
FEATURES•
Meets CDC guidelines for protection against TB.
•
NIOSH certified to have a filter efficiency of 95% or greateragainst non-oil based particulates.
•
Highest ASTM level of fluid resistance – 160 mm Hg.
•
Bacterial filter efficiency >99.9%.
•
Available in 5 color coded sizes.
•
Non-Latex respirator and straps.
•Dura-Mesh shell resists collapsing in regular use and evenunder hot and humid conditions.
•Contour molded nosebridge for extra comfort.
•Soft foam nose cushion for an easy and comfortable fit.
•Softspun inner lining is soft to the touch.
•Certified under 42CFR84.
•Meets heat and flame resistance in accordance with ANSI/ISEA
110-2009 Section 7.11.1.
•
100% PVC-Free for a greener alternative.

1511 Small

1512 Medium

POTENTIAL HEALTHCARE APPLICATIONSSurgery, hospitals and healthcare facilities, nurse/hospicecare, correctional facilities, police, fire and rescue teams, EMS/EMT, homeless shelters, occupational health facilities, school
healthcare offices. Note: For areas with oils present, you mustuse an R or P respirator.
Outekeep r Dura-M
clea s respir esh ®
ner looki atorng

Filter isprotected on bothsides by Dura-Meshsingr
Inne -Mesh nt collapidityae
Dur s prev nd hum
lehelp heat atabfromor ef®
n m cpu co fa
fts g is theoS nin stli gaina1513 Large

THE
DURA-MESH
DIFFERENCE

1517 Low Profile Nose Bridge(800) 421-0668 Ext. 550/512 | sales@moldex.com | www.moldex.com

1500 SERIES

Moldex-Metric, Inc.
Culver City, California, U.S.A.
+1 (800) 421-0668

N95 HEALTHCARE ANDSURGICAL MASKSPART #SIZESTRAP MATERIALDISPLAYQUANTITYCASE
QUANTITY

1510 N95

XS

Non-Latex

20/Box

8 Boxes1511 N95

S

Non-Latex

20/Box

8 Boxes1512 N95

M

Non-Latex

20/Box

8 Boxes1513 N95

L

Non-Latex

20/Box

8 Boxes1517 N95

LP

Non-Latex

20/Box

8 Boxes

XS = Extra Small | S = Small | M = Medium | L = Large | L/P = Low Profile Nose Bridge

ProductCodeSize

NSN Number

1511

S6515-01-496-0850

1512

M6515-01-496-0916

1513

L

6515-01-496-0918

WARNINGS TO USER
1.
Non-latex straps are used in the headbands of these respirators. However, individuals highly sensitive
to natural rubber latex may have an allergic reaction.
2.
This product does not eliminate the risk of contracting any disease or infection. Change immediately
if the respirator comes in contact with blood or fluids, including body fluids. OSHA has not set apermissible exposure level for airborne biohazards.
3.
This product has not been sold with warnings or use instructions for personnel involved in industrial
or related situations. If you are considering such uses, first contact the Moldex Technical Services
Department at +1 (310) 837-6500 or +1 (800) 421-0668, ext. 512.

DESCRIPTION
1.
The Moldex Healthcare N95 Particulate Respirator and Surgical Mask is designed to help provide
respiratory protection for the wearer. This product has been tested1 and certified by NIOSH as
an N95 respirator and as having a filter efficiency level of 95% or greater against particulate
aerosols free of oil. It is fluid resistant,2 disposable and may be worn in surgery or throughout
the hospital.
2.

This product contains no components made from natural rubber latex.

Tested in accordance with NIOSH 42 CFR 84
2
Passed ASTM F 1862 @ 160mm Hg
1

INTENDED USE
The various models of Moldex Type N95 Healthcare Particulate Respirators and Surgical Masks meet
CDC Guidelines for TB Exposure Control within healthcare facilities. These devices are also intended tobe worn by healthcare personnel during surgical procedures to protect both the patient and healthcare
personnel from the transfer of microorganisms, body fluids and particulate material. This productcontains no components made from natural rubber latex.
CONTRAINDICATIONS
1.
Not for use with beards, or other facial hair that prevents direct contact between the face andsealing surface of the respirator.
2.
Eyewear must not prevent direct contact between the face and sealing surface of the respirator.
3.
Not to be used on children.

THIS RESPIRATOR IS APPROVED ONLY IN THE FOLLOWING CONFIGURATION:TC-

PROTECTION1

84A-0013

N95

84A-5171

N95

RESPIRATOR

1510

1511

1512

1513

1517

CAUTIONS &LIMITATIONS2XXXXABCJMNOPXABCJMNOP

1. PROTECTION
N95 Particulate Filter (95% filter efficiency level) effective against particulate aerosols free of oil; time use restrictions
may apply
2. CAUTIONS AND LIMITATIONS
A- Not for use in atmospheres containing less than 19.5% oxygen.
B- Not for use in atmospheres immediately dangerous to life or health.
C- Do not exceed maximum use concentrations established by regulatory standards.
J- Failure to properly use and maintain this product could result in injury or death.
M- All approved respirators shall be selected, fitted, used, maintained in accordance with MSHA, OSHA, and otherapplicable regulations.
N- Never substitute, modify, add or omit parts. Use only exact replacement parts in the configurations as specified bythe manufacturer.
O- Refer to user’s instructions, and/or maintenance manuals for information on use and maintenance of these
respirators.
P- NIOSH does not evaluate respirators as surgical masks.

RESTRICTIONS:
1.
Before use, a written respiratory protection program in accordance with 29 CFR 1910.134 must beimplemented.
2.
This respirator does not supply oxygen and must not be used in atmospheres containing less than
19.5% oxygen.
3.
Prior to each use, carefully inspect the entire respirator, including filter media and strap attachment area
for tears and damage. Staple or strap attachment perforations do not effect NIOSH approval.
4.
If respirator is damaged, distorted, a proper fit cannot be obtained or breathing becomes difficult, leave
contaminated area as soon as possible and replace respirator.
5.
If the respirator comes in contact with blood or fluids, including body fluids, leave contaminated area assoon as possible and discard and replace the respirator.
6.
Do not alter, modify or abuse this respirator.
7.
Store unused respirators in box/bag in a clean, dry, non-contaminated area.
8.
Dispose of respirator according to your employer’s policy and local regulations.
9.
Do not reuse or store for reuse or hang around neck unless your employer specifically authorizes reuse.
10.
When used for surgical procedures, discard after every use.
11.
Use respirator before the “use by” expiration date printed on box/bag.
LIMITED WARRANTY IMPORTANT NOTICE TO PURCHASERThis limited warranty is made in lieu of the warranties of merchantability, fitness for particular purposes andall other warranties, express or implied. There are no other warranties which extend beyond the description on
the face hereof. The physical standards and specifications of Moldex will be met by products sold. Exclusive
Remedies: damages for the breach of this limited warranty are limited to the replacement of such quantity ofMoldex, products proved to be defectively manufactured. Except as provided above, Moldex shall not be liable
or responsible for any loss, damage, or liability, direct, indirect, incidental, special, or consequential, arising outof sale, use, or misuse, or the inability to use products by the user.

MOLDEX TECHNICAL ASSISTANCE
Moldex provides all of the technical assistance required to set up a hearing protection program.
For more information call +1 (800) 421-0668 or +1 (310) 837-6500 ext. 512 or email: tech@moldex.com
or visit www.moldex.com.

MOLDEX-METRIC, INC.

U.S. & CANADA CUSTOMER SERVICETEL: +1 (800) 421-0668, Ext. 550
10111 Jefferson Blvd.
Culver City, CA 90232
TEL: +1 (800) 421-0668 or +1 (310) 837-6500
FAX: +1 (310) 837-9563
E-mail: sales@moldex.com | www.moldex.com

MADE IN THE USA

1500-807 REV C 4/22