Product Information

PI"SD Series I.I."
X-ray Image Intensifier

E5796SD-P2A
Triple Input Field:
310/230/160mm (12”/ 9”/ 6”)
Output Image Diameter: 25mm

FEATURING: High Resolution: 46 Lp/cm
cd/m2
 High Conversion Factor : 34 μGy/s
 High Contrast Ratio
10% area contrast: 30

10mm dia. contrast

: 18



“SD Series I.I.” E5796SD-P2A is used for medical radiographic diagnosis.

This X-ray Image Intensifier is intended to be connected with IEC ClassⅠequipment.

“SD Series I.I.” E5796SD-P2A has been perfected by the integration of several aspects of advanced
technology；
The input window is of a thin metal plate, which has good X-ray transparency characteristics andreduces X-ray scatter.
It uses a thick input phosphor screen made with an extremely fine pillar shaped structure. This has
high X-ray absorption and very high conversion efficiency. In addition, there is a thin output
phosphor screen on a single thick-glass output window with an anti-reflective coating, offering very
high contrast for small details, very low structure noise and high MTF characteristics.
The E5796SD-P2A, employing these manufacturing techniques, provides outstanding performancein resolution, brightness, contrast and noise.
The E5796SD-P2A is supplied in a housing, which provides mu-metal magnetic shielding andradiation shielding.

★The information contained herein is presented only as a guide for the application of our products. No responsibility is assumed by Canon Electron Tubes& Devices Co., Ltd. (CETD) for any infringements of patents or other rights of the third parties which may result from its use.
No license is granted by implication or otherwise under any patent or patent rights of CETD or others.
★The information contained herein may be changed without prior notice. It is therefore, advisable to contact to CETD before processing with the design ofequipment incorporating this product.

E5796SD-P2A

PERFORMANCE DATA
Item
Central Resolution
Conversion Factor (Gx)
10% area
10 mm dia.
DQE (IEC standard)
Contrast Ratio

N mode4634300
301865

M1 mode50-

M2 mode56-

Lp/cm(cd/m2)/(μGy/s)(cd/m2)/(mR/s)%GENERAL DATA MECHANICAL:
Item
Overall Length
Maximum Diameter
Nominal Entrance Field Size
N (12″) mode
M1 (9″) modeM2 (6″) mode
Output Image Diameter
Weight (approx.)
Dimensional Outline

434.5±5 mm
380±2 mm
310 mm min.
230±5 mm
160±5 mm
25±0.5 mm
31 kg
Refer to page 4

 POWER SUPPLY:
ItemHigh Voltage Power Supply
Input Voltage

24±1 Vdc

 ENVIRONMENTAL:
Item
Temperature
Humidity
Pressure

Under delivery and stock-15 to 45 ℃
10 to 90 %50 to 106 kPa-2-Under operating
5 to 40 ℃
30 to 85 %70 to 106 kPa

typ.
typ.
typ.
typ.
typ.
typ.

E5796SD-P2A

ELECTRICAL WIRING AND CONNECTINGX-ray image intensifier is operated with 24Vdc input voltage applied to the high voltage power supply.
Wiring and connecting are shown as follows.

DIN connector

Pin assignment of DIN Connector* Switching of entrance field size
N mode .......................... 4, 5 OpenM1 mode ........................ 1-4 Close (apply DC24V to Pin No.4)
M2 mode ........................ 1-5 Close (apply DC24V to Pin No.5)

1. DIN connector ............................................................. one connector attached with each product2. GND cable ............................................................................................................................. 3m×1-3-E5796SD-P2A

DIMENSIONAL OUTLINE
Unit: mm

Ⓐ Ⓑ : Screw holes for fixing to apparatus

Marketing Engineering Group, Sales Department
1385, Shimoishigami, Otawara-shi, Tochigi 324-8550, Japan
Tel: +81-287-26-6666
Fax: +81-287-26-6060
https://etd.canon

PE-E5796SD-P2A 2018-11