Issued 5.15.2017
No.

Ｔ315A04Be

B

Remote sensor system
8 signals of thermocouple speciﬁcations
RS080series

Thank you so much for purchasing our products.
Before using this Processor, read this manual carefully and operate properly, paying attention
to the safety aspects.

Remote system /8 signals thermocouple speciﬁcations■ Index ■1.Product summary
1.1

System conﬁguration

・・・・・・・・・・・・・・・・・・・・・・・・3

2. Transmission ampliﬁer speciﬁcations
2.1.

Product speciﬁcation ・・・・・・・・・・・・・・・・・・・・・・・・・4

2.2.

Accessible temperature sensor ・・・・・・・・・・・・・・・・・・・・4

2.3.

LED indication ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・4

3. Transmission head speciﬁcations
3.1.

Compact shape

3.2.

Ring shape・・・・・・・・・・・・・・・・・・・・・・・・・5
・・・・・・・・・・・・・・・・・・・・・・・・・・5

4. Output head speciﬁcations
4.1.

Compact shape・・・・・・・・・・・・・・・・・・・・・・・・・・6

4.2.

Ring shape・・・・・・・・・・・・・・・・・・・・・・・・・・6

5. Current output ampliﬁer
5.1.

Product speciﬁcation ・・・・・・・・・・・・・・・・・・・・・・・・・7

5.2.

Output electric current level ・・・・・・・・・・・・・・・・・・・・・・7

5.3.

LED indication ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・7

6.

Wiring6.1. Wiring of a transmission ampliﬁer and the transmission head ・・・・・・・8
6.2. Wiring of a transmission ampliﬁer and the transmission head ・・・・・・・8
6.3. Wiring of the output head ・・・・・・・・・・・・・・・・・・・・・・・・・96.4. Connection of current output ampliﬁer

・・・・・・・・・・・・・・・・・9

7.Installation
7.1. Neighborhood metal and mutual interference and transmission domain ﬁgure ・・・・・・・10
8.Protocol
8.1.

Communication Setting ・・・・・・・・・・・・・・・・・・・・・・・・11

8.2.

Regarding a command ・・・・・・・・・・・・・・・・・・・・・・・・・11

8.2.1.

Command for data requirements ・・・・・・・・・・・・・・・・・・・・11

8.2.2.

Command for stopping automatic answering mode ・・・・・・・・・・11

8.2.3.

Command for starting automatic answering mode

8.3.

List of error codes (ASCII notation) ・・・・・・・・・・・・・・・・・・11

9.

・・・・・・・・・・11

Setting method of the transmission ampliﬁer
9.1.

Wiring to a transmission ampliﬁer ・・・・・・・・・・・・・・・・・・・12

9.2.

Setting of the transmission ampliﬁer by the application software ・・・12

10.

Setting of the output head ・・・・・・・・・・・・・・・・・・・・・・・・・・

10.1.

Wiring of the output head ・・・・・・・・・・・・・・・・・・・・・・・13

10.2.

Setting of the output head by the application software ・・・・・・・・13

Request for use on※ Speciﬁcations subject to change without notice.
If there is a signal of notice about the contents of this document, hope you'll give me your contact us, thank you.
-2-

Remote system /8 signals thermocouple speciﬁcations

1.Product summary
This product feeds from an output head to a transmission head. And feeds to the bottom apparatus using the electromagnetic induction technology.
In addition, it is transmitted to an output head by wireless via a transmission ampliﬁer to a transmission head, and temperature measuring of a lower temperature sensor is sent to the higher apparatus. Temperature measuring signal sent by an output head is the serial
signal (RS-232C). It is possible to output 4...20mA currency by putting output ampliﬁer.
The temperature sensor which can measure is a thermocouple (8 signals), a resistance thermometer (2 signals), thermistor (2 signals).
Possible to presence the detection of guide wire malfunction of the sensor by setting attached application. Also variety temperature
sensors and the number of the temperature sensors.
In addition, at the time of the current output ampliﬁer use, it is possible to set temperature in the 4mA output and temperature in the
20mA output.

1.1 System conﬁguration
The system constitution of the product is as follows.
Transmission
ampliﬁer

Temperature
sensorTransmitting
head

Output Head

Compact shapeCompact shape

RSH080T-422-CN

RSH080E-422R-CN

ring shapering shape

Current output
Ampliﬁer

PowersupplyPower

Thermocouple
PLC
Resistance

Current

RS801E-234E

thermometer

Signal

PLC

RS080T-233

typePower supply

Thermistor
RSH080T-R01-CN

RSH080E-R01R-CN

Transmitter

PC

Serial communication

etc

Output sensor

A transmission head and the output head have two kinds of the compact shape and ring geometry.
Please choose upon the usage of the environment.
DeviceProduct type code

Description of each part

Transmission ampliﬁer

RS080T-233

Able to connect a temperature sensor to up to 8 signals.

Transmitting
Compact shape RSH080T-422-CN
head

Ring shape

RSH080T-R01-CN

Receiving wireless feeding from an output head and supply
electricity to a lower device,

Output Head Compact shape RSH080E-422R-CN Wireless feeding to a transmission head and transmit the

Ring shape
Current output ampliﬁer

temperature signal which has been sent to a higher apparaRSH080E-R01R-CN tus in RS-232C
RS801E-234E

A serial signal(RS-232C) which has been sent by output
head converts into a current signal (4..20mA)-3-

Remote system /8 signals thermocouple speciﬁcations

2. Transmission ampliﬁer speciﬁcations
2.1.

Products speciﬁcation
Type code
Application trans- Corner shape

mission head

ȭ









ring shape

Operating temperature / humidity
Storage temperature / humidity

Output interface

RS-232C Connects to the PC, and the numberof sensors and the diﬀerent setting are possible

Applicable sensor

Thermocouple (up to 8 signal)
Resistance bulb (up to 2 signal)
Thermistor (up to 2 signals)
Thermocou- 6Hz / 1ch without the disconnection detection
ple 4.2Hz / 1ch with the disconnection detection
Resistance 6Hz / 1ch
thermometer
Thermistor 6Hz / 1ch± 0.1℃

Output rate

Accuracy

Compensated error of the cold junction 0.2 degrees Celsius (at the time of thermocouple use)


2.2.

RS080T-233
RSH080T-422-CN
RSH080T-R01-CN
0...70℃ / 35...90%RH
0...80℃ / 35...90%RH

Resolution
Protection class
Case material
Standard
Weight0.01℃
Without protectionABS
CE acquisition ﬁnished
80g

Accessible temperature sensor

The temperature sensors that this product is accessible are as follows■ Thermocouple
type
J
K
E
N
R
S
T
B

Lower limit■ Resistance thermometer
Maximum limit

Temperature（℃） Temperature（℃）

-210
-265
-265
-265
-50
-50
-265
40

type

Lower limit

RTD PT-10
RTD PT-50
RTD PT-100
RTD PT-200
RTD PT-500
RTD PT-1000
RTD 1000
RTD NI-120

1200
1372
1000
1300
1768
1768
400
1820

Maximum limit

Temperature（℃） Temperature（℃）

-200

850

Thermistor
Lower limit

type

Maximum limit

Temperature（℃） Temperature（℃）

44004/44033
44005/44030
44007/44034
44006/44031
44008/44032
YSI 400
Spectrum 1003K

-40

150

-80
-50

250
125

The transmission ampliﬁer connects a transmission ampliﬁer via RS232C and operates setting by application (Please see page12)
Parts
Contents
Category of the sensors Type of the temperature sensor to connect*1
Type of the sensors
Type of the temperature sensors
Numbers of the sensors The number of sensors to connect
The open detection
Detection of the sensor that is unconnected (disconnection) *2
min.
Temperature in 4mA
max.
Temperature in 20mA

Factory setting
Thermocouple
J type
8
Yes
0℃
200℃

*1 Cannot be use in combination with a diﬀerent in the kind sensor.
*2 When choosing a resistance thermometer and thermistor, it becomes the disconnection detection forcibly.

2.3.LED (indication)
Two LED is put on a transmis- ■ LED1
sion ampliﬁer and shows a kind Color Temperature senand the type of a connectedsor
temperature sensor with a col- Green Thermocoupleor.
Blue ResistanceRed■ LED2Color
Thermocouple
Light blue J type
Green K type
Violet E type
Yellow N type
R typeOFF
S typeRed
Ｔ typeBlue
White B typethermometer type
Thermistor-4-

Resistance thermometer
PT10
PT50
PT100
PT200
PT500
PT1000
RTD1000
RTD NI-120

Thermistor
44004/44033
44007/44030
44007/44034
44006/44031
44008/44032
YSI 400
Spectrum 1003k

Remote system /8 signals thermocouple speciﬁcations

3. Transmission head speciﬁcations
Compact shape
Type code
RSH080T-422-CN
Rating transmission distance 0...3mm
Center oﬀset± 3mm
Operating temperature
0...70℃ / 35...90%RH/ humidity
Storage temperature
0...80℃ / 35...90%RH/ humidity
Protection class
IP67
Case material
ABS
Connection
M8 connector 3, Pin female
Standard
CE acquisition ﬁnished
Weight
110g

‫ڧ‬

‫ڧ‬

3.1.

ȭ
ȭ



Pin
1
3
4

㸯







3.2.

Color
Brown
Blue
Black

Descriptions
Power
Gound
Signal input

Ring shape

㸲㸫


ȭ


ȭ

inner diameter)ȭ (Sleeveྲྀ௜㺛㺶㺎㺪㺼ෆᚄType code
RSH080T-R01-CN
Rating transmission dis- 0...6mm
tance
Center oﬀset± 6mm
Operating temperature / 0...50℃ / 35...90%RH
humidity
Storage temperature / hu- 0...70℃ / 35...90%RH
midityProtection class
IP67
Case material
PUR
Connection
M8 connector 3, Pin female
Standard
The CE non-acquisition



r


ȭ






㸯
Pin
1
3
4









-5-

Color
Brown
Blue
Black

Descriptions
Power
GroundSignal input

Remote system /8 signals thermocouple speciﬁcations

4. Output head speciﬁcations
4.1.

Compact shape
Type code
Power supply
Current consumption

‫ڧ‬

‫ڧ‬

Operating temperature/ humidity

ȭ
ȭ


,1=21(/('







4.2.










RSH080E-422R-CN
24V DC ± 10%（incl. ripple）≦ 100mA
0...70℃ / 35...90%RH

Storage temperature / humidity

0...80℃ / 35...90%RH

LED (indication)

Orange: In zone
Lights-out: Application transmission without headInterfaceProtection class
Case material
Connection
Standard
Weight

RS-232C
IP67
ABS
M8 connector 6 pin male
CE acquisition ﬁnished
110g

Pin Color Descriptions
1 Brown Power supply +24V
2 White Transmitting (TX)
3 Black Receiving (RX)4Blue Power supply ground (GND)
5 Green Signal ground (SG)RTS RS232C
In zone output
6
Red H…Transmission head
L: Transmission without
head

Ring shape



ȭ




ȭ

inner diameter)ȭ (Sleeveྲྀ௜㺛㺶㺎㺪㺼ෆᚄType code
Power supply
Current consumption
Operating temperature/ humidity



r


ȭ
















Storage temperature / humidity

0...70℃ / 35...90%RH

LED (indication)

Orange : In zone
Lights-out: Application transmission
without headInterfaceProtection class
Case material
Standard
Weight

RS-232C
IP67
PUR
The CE non-acquisition
470g






-6-

RSH080E-R01R-CN
24V DC ± 10%（incl. ripple）≦ 100mA
0...50℃ / 35...90%RH



Pin Color Descriptions
1 Brown Power supply +24V
2 White Transmitting (TX)
3 Black Receiving (RX)4Blue Power supply ground (GND)
5 Green Signal ground (SG)RTS RS232C
In zone output
6
Red
H…With transmission head
L: Transmission without headRemote system /8 signals thermocouple speciﬁcations

5. Current output ampliﬁer
5.1.

Speciﬁcation
Type code



5.2.



ȭ










Applicable Trans- Corner typemitter head

Ring shape
LED (indication)

RS801E-234E
RSH080E-422R-CN
RSH080E-R01R-CN
Blue: Power supply ON, before initialization
Green: Transmission without head
Red: Transmission with head (in zone)

Power supply
Current consumption
Resolution
Accuracy

24V DC ± 10%（incl. ripple）≦ 250mA (in a current output maximum)
0.002%FS± 0.5%
Operating temperature / humidity 0...50℃ / 35...90%RH
Storage temperature / humidity 0...70℃ / 35...90%RH
Load resistance400 Ω or lessProtection class
Without protection
Case material
ABSStandard
CE acquisition ﬁnished
Weight110g

Output electric current level

The current value output and current output ampliﬁer are as follows.
Current value Contents
0mA
A transmission head is not corresponding
4mA
The minimum temperature set to a transmission ampliﬁer
20mA
The maximum temperature set to a transmission ampliﬁer
Disconnection of the sensor
21mA(only when setting is on for checking disconnection)
Conversion type from a current value to temperature
Temperature ( ℃ )=( max temp. --min
min temp.) x(Current value − 4mA）
16mA

+min.temp.

5.3.LED (indication)
LED indicates the condition of
the current output ampliﬁer.

Color Contents
Blue Turns on when power is on.Turns oﬀ after the initialization completed.
Green At the time of transmission head non-facing
Red Inzone-7-Remote system /8 signals thermocouple speciﬁcations

6.About the wiring
6.1. Wiring of a transmission ampliﬁer and the transmission headTransmission ampliﬁer
Brown

VCC

Black

TX

Transmitting
head

RX
Blue

GND

※ The connection cable uses the following cable
03M-M08S01-01（Straight）
03M-M08R01-01（Angle）
The model end is the length of the cable（01=1m)

Connection of the transmission side
Terminal stand of the transmission ampliﬁer
IN1+
IN1 −
IN2+
IN2 −
IN3+
IN3 −
IN4+
IN4 −
IN5+
IN5 −
IN6+
IN6 −
IN7+
IN7 −

Sensor 1 in put ＋
Sensor 1 in put‐
Sensor 2 in put ＋
Sensor 2 in put‐
Sensor 2 in put ＋
Sensor 3 in put‐
Sensor 4 in put ＋
Sensor 4 in put‐
Sensor 5 in put ＋
Sensor 5 in put‐
Sensor 6 in put ＋
Sensor 6 in put Sensor 7 n put ＋
Sensor 7 in put -

IN8+
IN8 −
VCC
TX
RX
GND

Output headpin assignment

Sensor 8 in put ＋
Sensor 8 in put Power input
TX RS232C transmits
RX RS232C receives
Power supply ground

No of

Cable

the pin
1
3
4

color
Brown
Blue
Black

Descriptions
The VOUT power supply output

GND Ground
SN Signal input

6.2. Wiring of a transmission ampliﬁer and the transmission head
■ Thermocouple connection■ Resistance thermometer connection

Transmission ampliﬁer
Thermocouple 1
Thermocouple 2

Thermocouple 8

+
+
-

+
-

IN1+
IN1 −
IN2+
IN2 −

・
・
・
IN8+
IN8 −

■ Thermistor connection

Transmission ampliﬁer
Resistancethermometer 1

Resistancethermometer 2

When you connect a thermocouple, placethem from IN1, and then connected.
When there is a gap between the interval
it can cause the disconnection error.

-8-

Transmission ampliﬁer

IN1+
IN2+
IN3+

Thermistor 1

IN1+
IN2+
IN3 −

IN6+
IN7+
IN8+

Thermistor 2

IN6+
IN7+
IN8 −

Remote system /8 signals thermocouple speciﬁcations

6.3.

Wiring of the output head

Up to Max. 5m

Brown

24V

Black

GND

External power supply

Output Head

※ The connection cable uses the following cable
06F-M08S01-02（Straight）
06F-M08R01-02（Angle）
The model end is the length of the cable（02=2m)

PLC

D-sub9 pin

White RX2Black TX

3

Green SG

5

１

5

6

9

Red CTS8

Output headpin assignment
No of

Cable

the pin
1
2
3
4
5
6

color
Brown
White
Black
Blue
Green
Red

Descriptions＊ The cable on the host device side is loose wires.
Please contact us if you need a connector cablewith D-sub 9 pins

+24V Power input
TX RS232C transmits
RX RS232C receives
Power supply ground (GND)
Signal ground (SG)RTS RS232C in one output

When digital output, connects to PC.

6.4. Connection of current output ampliﬁer
Output ampliﬁer

To Max. 5m

GND
Black
White

The data output (an output head and connection)
The data input (an output head and connection)
The data output (an output head and connection)
Grand (Output head and connection)

COM
OUT1
OUT2
OUT3
OUT4
OUT5
OUT6
OUT7
OUT8
COM
VCC
GND

Current output ground
Current output 1
Current output 2
Current output 3
Current output 4
Current output 5
Current output 6
Current output 7
Current output 8
Current output ground
Power input（+24V）
Power supply ground

TX
RX
COM

Brown
Blue

Current output ampliﬁer terminal stand
TX
RX
VOUT
SGND

24V
GND External power supply

Output Head

※ The connection cable uses the following cable
06F-M08S01-02（Straight）
06F-M08R01-02（Angle）
The model end is the length of the cable（02=2m)

VCC

VOUT
OUT1

Load resistance

OUT2

Load resistance

OUT3

Load resistance

SGND

・
・
・
OUT8

-9-

Load resistanceRemote system /8 signals thermocouple speciﬁcations

7.Installation
7.1. Neighborhood metal and mutual interference and transmission domain ﬁgure
In order to avoid inﬂuence of surrounding metal, or to avoid mutual inﬂuence between parallel-mounted sensors,
keep the minimum free zone as described below.

■ When it is compact type■ Surround Metal■ When it is ring type

Parallel Setting■ Surround Metal

Parallel Setting

C
A

A

C
B

B

Type code
RSH080T-422-CN
RSH080E-422R-CN

A*

B

C

22

25

200

Y

6
Y

4
X

3

X:Center offset(mm)
Y:Operating distance (mm)2Operating area

1
8

6

4

2

0

2

4

6

C

22

35

200

mm
18
16
14
12
10
8
6
4
2

7
5

B

(mm)

mm

8

A

(mm)

* Possible that only one side, the metal contacts.

Y

Type code
RSH080T-R01-CN
RSH080E-R01R-CN

X-16-14-12-10 -8 -6 -4 -2 0 2 4 6 810 121416

X
8- 10 -YX

X:Center offset(mm)
Y:Operating distance (mm)Operating area

Remote system /8 signals thermocouple speciﬁcations

8.Protocol
Communication protocol speciﬁcations when transmitting to a command from a higher apparatus to the output head.
Please set the communication baud rate by attached application before communicating.

8.1.

Communication Setting

Baud rate
Stop bits
Parity115200、57600、38400、19200、9600、4800
１、２

mode:

Command mode, auto answer mode

Even number, Odd number,None*There are two movement modes to this product.
① Command mode
Transmit data acquisition command and receiving data.
It is necessary for the superior apparatus to transmit a command

8.2.

The bold-face is factory time of shipment② Auto answer mode
When a transmission head faces, it is the mode which data automaticallyanswer. The higher apparatus does not have to transmit a command and is
answered data whenever temperature changes

About a command

The commands are as follows.
CommandD
AE
AS

Contents
Temperature data demand command
Command for stopping automatic answering mode
Command for starting automatic answering mode

The bold-face writes ASCII, and the small characters are a hexadecimal notation.

8.2.1.

Command for data requirements

Read the temperature data of the channel. Please make sure to stop an auto-answer mode then transmit after
making it to a interactive mode.
CommandD・response

CH number【Example】

D CH number Special number Six columns of temperature data

the reception temperature data were 123.45 degrees Celsius.

【 Normal 】

Two columns of four columns of integer part + decimals parts+ or 【Abnormality】

8.2.2.

Transmitted a temperature data acquisition command of ch2, andCommandD

2

Response

D

2

+

0

1

2

3

NAK Error code

Command for stopping automatic answering mode

Stop an auto-answer mode and make it to interactive mode.
・commandAE・response【 Normal 】

ACK【Abnormality】

NAK Error code

8.2.3.

Command for starting automatic answering mode

Make temperature data an automatic-answering mode at the time of transmission department facing.
・commandAS・response【 Normal 】

ACK【Abnormality】

NAK Error code

8.3.

List of error codes (ASCII notation)

Error code
1
2
6
7
8

Transmission department does not exist
Disconnection error
Interface error
Format error
Parameter error

There is no reply from a transmission head.
A sensor is not connected
The communication setting including the baud rate includes a mistake.
Failure on command procedures
Failure on command procedures- 11 -

45

Remote system /8 signals thermocouple speciﬁcations

9.

Setting method of the transmission ampliﬁer

9.1.

Wiring to a transmission ampliﬁer

Connect a higher apparatus to a transmission ampliﬁer via an RS232C cable and set a movement condition
using attached application.
+24V
GND

Transmission ampliﬁer

D-sub 9-pin
VCC
RX

TX

TX

RX

SG

GND

9.2.

2
3
5

Setting of the transmission ampliﬁer by the application software① Start SensorSet.exe.
② Choose connected COM port number among COM Port.
③ After pushing Start, push Read. The setting of the transmission ampliﬁer is displayed.
④ Change a parameter, and setting is written when push Write.
①■ Parameter

②

③

④

Sensor Category

Category

Pick the sensor kind to connect.

of the sen-(thermocouple, resistance bulb,

Sensor Typesors
Type of thethermistor)
Pick the type of the sensor.

Number of sensors

sensors
Numbers of

Pick the sensor kind to connect.

Open detection

the sensors
The open

Set the operation for detecting un-

detectionconnected sensors.

Measurement temperature■ Measurement temperature
The measurement temperature is eﬀective at thetime of current output ampliﬁer connection.
min
max.

Set temperature to output 4mA.
Set temperature to output 20mA.

※ It is eﬀective only when connect a current
conversion ampliﬁer- 12 -

Remote system /8 signals thermocouple speciﬁcations

10.

Setting of the output head

10.1.

Wiring of the output head

Please see page 7「wiring of the output head」

10.2. Setting of the output head by the application software① Start attached application Temp8ch.
② Click COM Port and select a connected communication port.
③ Scans the communication conditions such as baud rates when push the Scanning button④ when push the Property button, RS232 communication format and a parameter setting window display.
choose a baud rate, stop bit, parity, a movement mode and then press a write down button.
⑤ Select the number of the sensor which you want to begin to read by ch choice, and temperature data are displayed whenpushing the temperature data demand button.
⑥ Polling is started when pushing the automatic start button, and temperature indication is automatically displayed.

①

②

③

④

⑤

⑥

Temperagure data

Temperagure data request

Automatically request

Stop the automatic request

RS232C communication format

Baud rate

Operating mode

Write down

Stop bit

buttonParity- 13 -

Remote system /8 signals thermocouple speciﬁcations- 14 -

Remote system /8 signals thermocouple speciﬁcations- 15 -

Wireless Power Supply by

Mail ：b-plus-usa@b-plus-kk.com
Web ：http://www.b-plus-kk.com* Infor may change the mention contents such as speciﬁcations without a notice.
Thank you for understanding