FS2000

DIGITAL INDICATOR WITH GRAPHIC DISPLAY / TOUCH PANEL(SD CARD SLOT & HIGH SAMPLING RATE)

The best solution for OK/NOK judgment of press fitting and caulking application !!
High responsiveness of 5 kHz to fully utilize the performance of Super Cell !!
A fluctuation of force is shown as a waveform!!
‫ ە‬Two-dimensional OK/NOK judgement can be performed‫ ە‬25000 times/sec. high-speed processing‫ ە‬Variety of interfaces

with a load cell and displacement sensor.
‫ ە‬Analog monitor output

USB / CC-Link / DeviceNet / EtherNet/IP / Ethernet / PROFINET IO

Voltage output is proportionate to the input signal making‫ ە‬4.3-inch color LCD module & touch panel

the recording on recorder convenient.

Operation can be effortlessly performed by a direct touch

Approx. 2 V per 1 mV/V strain gauge input

on the touch panel.

Comparison & hold function by waveform display

Improved usability‫ ڦ‬Waveform comparison
This function compares the actual measurement waveform
against the setup High/Low limit waveforms and will give out anNOK judgment when any of the point exceeded the preset
High/Low limit waveforms.

4.3 inch wide display provides excellent visibility.
Main display configuration can be selected to keep it assimple as possible by eliminating unnecessary information.

‫ ڦ‬Multi-point judgment
OK/NOK judgment can be performed on multi points in one
process. (e.g. The start point and end point of press fitting can bejudged respectively.) (Max. 5 points)<Multi-point judgment>
Peak hold
Inflection point hold+Peak holdPeak(Multiple holds can be adjusted
in the same range.)<Waveformcomparison>

OK judgment

PeakHi limit setting
Lo limit setting

Inflection pointX axis:
time ordisplacement

Y axis: load

Enlarged numerical display

Selectable waveform reference
Judgement is possible based on press point of press machine
Load

Load

010 mmFront reference

Displacement

Measurement point starts from left

unlPULXE-10 mm0Displacement

Enlarged waveform display

Back reference

Measurement point starts from rightwww.unipulse.tokyo/en/Trend display is helpful for preventive maintenance

Changed setting items are highlighted!

Trend of the zero-point shift and hold values can be monitored
to find any irregularities for preventing breakdown of machines.

Master and current set values are listed up for checking thechanged setting items easily.
Set values can be edited directly on the list without going into
each setting menu.
* Except for waveform comparison settings

Trend display

List display

Saved measured data (waveform) on SD card can be displayed afterwards

Example of use
FS2000Measured data and set values can be saved in
the SD card. Data can be converted to CSV
format easily for editing with Microsoft Excel.

SD cardDisplacement sensor
USB, Ethernet

Load cell
I/OPC

CC-Link, DeviceNet, EtherNet/IP,Ethernet, PROFINET IO

Press machine

PLC

FS2000-HYS Hysteresis specifications

Hysteresis specifications

Standard Can see going waveform

Can see outgoing andreturn waveform‫ ە‬Can choose comparison method!
<Differential waveform comparison>
Judgment of OK/NOK by the difference between
going and returning<Standard waveform comparison>
OK/NOK judgment for going and returning waveform.

kN
40.00

kN
40.00

Big difference between
going and returning

Going waveform
is OK

Returning waveform
is NG-10.00-10.000.00

50.00 mm

Recommended for below usage:- Expansion and contraction of the spring- Rotating the steering wheel clockwise,
counterclockwise, etc.

0.00

50.00 mm

Recommended for below usage:- Torque hinges used to open and close doors- Shock absorbers that absorb the impact of tires, etc.

FS2000-MHP Multi hold point specifications
Standard
OK/NOK judgement up to 5 points
Up to 15 point detection
sections can be set

Waveform comparison
function is also available

Multi hold point specifications
2.12.MXGPHQWXSWRSRLQWV

unlPULXE

www.unipulse.tokyo/en/Sensorinput(Standard)

Sensor input for load (Fixed as strain gauge input) (6-wire)
Excitation voltage DC 2.5, 5, 10 V±10% (depending on settings) Output current: Within 30 mA
Signal input range 㸫2.0 to 㸩2.0 mV/V
AccuracyNon-linearity: Within 0.02% FS±1 digit (at 2.0 mV/V input)
Zero drift:
Within 0.1 ȣV/Υ RTI
Gain drift:
Within 15 ppm/ΥSelectable from 10 Hz to 10 kHz (-6 dB/oct.) (at A/D converter speed 25000 times/sec.)
Low-pass filter
Selectable from 2 Hz to 2 kHz (-6 dB/oct.) (at A/D converter speed 5000 times/sec.)

Interface

USB:CCL:ODN:

A/D converter

Sensorinput
Multisensor
input(Option:[MLT] or[MLT2])

Speed:
Selectable from 25000 times/sec., 5000 times/sec.
24 bit (binary) Effective resolution: Approx. 1/20000 against 2.0 mV/V
Resolution:
Sensor input for displacement (Pulse input: Line driver)Max. input frequency 1 MHz
Internal count range Approx. 1,000,000
Adaptable encoder Output: Incremental type 2-phase output (A/B-phase signal output)
Also capable of single-phase output(A-phase input used. All pulses are counted as in the plus direction.)
Output stage circuit specification: Line driver (Based on RS-422)
Sensor input for load (Strain gauge) (6-wire) … Same as standard
Sensor input for displacement (Pulse input: open collector) ... Other than output circuit, spec is standard [MLT]
Output stage circuit specification: Open collector
Sensor input for displacement (Pulse input: line driver) … Same as standard [MLT2]
Voltage input
Signal input range 㸫10 to 㸩10 V
Absolute max. rating ±15 V

Input signal (16)

Input impedance
Accuracy

Approx. 1 MΩ or more
Non-linearity: Within 0.02% FS±1 digit (at 10 V input)
Zero drift:
Within 0.2 mV/Υ RTI
Gain drift:
Within 0.01%/Υ
Selectable from 10 Hz to 10 kHz (-6 dB/oct.) (at A/D converter speed 25000 times/sec.)
Selectable from 2 Hz to 2 kHz (-6 dB/oct.) (at A/D converter speed 5000 times/sec.)

Low-pass filter
A/D converter

Speed:
Selectable from 25000 times/sec., 5000 times/sec.
Resolution: 24 bit (binary) Effective resolution: Approx. 1/20000 against 10 V
AnalogOutput levelApprox. 2 V per 1 mV/V input
voltage output
Load resistance: 2 kΩ or more
Display
4.3 inch TFT color LCD module, Display area: 95.0(W) × 53.9(H) mm, Dot configuration: 480 × 272 dot
Display frequency
Fixed at 3 times/sec.
Comp.
Multi point comparison mode: 16 ch (set values)&Capable of judging up to 5 hold points at the same time.
judge.
Sample, Peak, Bottom, P-P, Relative Maximum, Relative Minimum,
function
Inflection Point (A,B,C,D), Average, End displacement
Waveform comparison mode: 16 ch (setting values can be stored)
Compares the actually measured waveform against the preset HI/LO waveforms.
The overall measured waveform will be compared against the preset HI/LO and if anyof its points exceeds the preset waveform, then the measured waveform will be NOK.
Hysteresis
Multi-point comparison and waveform comparison are possible by measuring going/returning
specifications with one waveform. (Can choose go/return difference comparison)
Number of drawing points: 1000 points for going, 1000 points for returning
Multi hold point Multi hold: 15 points
specifications Sampling speed: 5000 Hz
Showing the trend of measurement data to help finding irregularities at early stage.
Preventive
Trend display
maintenance Statistics
supportScreen captureWork name edit
Setting list display
User management
External
Output signal (16)signalUsing the latest 10000 measured data
Displaying number of measurement, OK, NOK
Saves screen capture data as bmp data.
Work name can be edited and displayed for each Work No.
Changed setting items comparing to master set values are highlighted.
Login ID and Password
Point judgment (load, displacement)/ Load overload/
Measurement complete/ Waveform comparison judgment/
Load & displacement OK/ CPU OK/ SD card OK/ Timing output 1,2
Output Type: Sink type/ source type selectable.
(Source Type is option: [ISC])
Output transistor ON at signal ON.
To connect an input unit like a PLC, connect plus
common for sink type, and minus common for source type.
Rated voltage: 30 V, Rated current: 30 mA* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,fluorescent display tubes and LCDs due to manufacturing process or production lots.

Load digital zero/ Displacement adjustment/ Measurement start/
Measurement end/ HOLD1 to 5/ Reset/
Forcibly light up the backlight/ Touch panel lock/ Work change
Input type: Plus common/ Minus common selectable.
(Minus common is option: [ISC])
To connect a transistor, connect NPN output type (sink type) for
plus common and PNP output type (source type) for minus common.
USB interface
EIP:
EtherNet/IP interface (option)
CC-Link interface (option)
ETN:
Ethernet interface (option)
DeviceNet interface (option)
PRT:
PROFINET IO interface (option)*Only one option can be installed

Option

ISC: I/O Source board, MLT: Multi sensor input, MLT2: Multi sensor input 2Special

FS2000-HYS:

FS2000Specifications

Special option which records and judge a reverse waveform

option(Hysteresis specifications)
FS2000-MHP:

Special opeton which enables to detect hold points up to 15(Multi hold point specifications)
GeneralPower supply voltage
DC 24 V (±15%)specifications Power consumption
6 W typ.
Operating conditions
Operation temperature: 㸫10 to 㸩40Υ
Storage temperature: 㸫20 to 㸩60Υ
Humidity: 85% RH or less (non-condensing)
Dimension
132(W) × 98(H) × 110(D) mm (Not including projections)
WeightApprox. 1.0 kg
CC-Link connector
Attachments I/O connector (with cover) …1(when CC-Link option is selected) ………1
Analog connector ……………1
DeviceNet connector
Operating tool ………………1(when DeviceNet option is selected) ……1SD card 1 GByte ……………1
Operation manual ……………1
Optional
accessories

CN36:

I/O connector (with cover)(Same as the attachment)
CN71: CC-Link connector
CN72: Double row connector for CC-Link
CN77: Analog connector(Same as the attachment)
CND01: DeviceNet connector

TSU03:SD1G:

DC lighting surge unit
SD card 1 GByte(Same as the attachment)
SD2G:
SD card 2 GByte
SD16G:
SD card 16 GByte
SD32G:
SD card 32 GByte
CA81-USB: USB cable (A-miniB type) 1.8 m

CE marking EMC directive EN61326-1
certification

Structure of product codeFS2000
ձձ Standard unit

‫ڧ‬ղ

‫ڧ‬
ճ‫ڧ‬
մ

ղ I/O output

մ Interface

SignStandard
ISC

Sign
Standard

Output type
Sink type (NPN output)
Source type (PNP output)

InterfaceUSB↓ One optional interface can be added
in addition to the standard interface.

ճ Sensor input

CCL
ODN
EIPETN
PRTSign
Output type
Standard Strain gauge,Line driver
MLT orStrain gauge
MLT2Open collector (MLT only)
1*Line driver (MLT2 only)
Voltage (Load or displacement)

CC-LinkDeviceNet
EtherNet/IP
Ethernet *2
PROFINET IO*2 When choose ETN option,
USB interface is not included.

*1 When choose MLT option, ULE-50 is not available to use.

Combination table
X axis

Y axisTime
Line driverLine driver
Time
Open collector
Open collector

Strain gauge
Strain gauge
Voltage (Load)
Voltage (Load)
Strain gauge
Voltage (Load)

Voltage (Displacement)

Strain gauge

Standard○○
×××××

MLT○××○○○○

MLT2○○○○××○

External dimension(Rear)

Unit: mm

Panel cutout size

SENSOR1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 1691

DC IN

92 +10(Side)

98(Front)

Panel thickness
1.6 to 3.2 mm

CONTROLUSBMADE IN JAPAN

132

10

100

SER. NO.

124

125 +10A digital contact sensor designed for FS2000
Digital contact sensor

ULE-50You can perform OK/NOK judgment
nt curve.
with a Force vs Displacement

Wide measuring range & high-accuracy
Measuring range: 50 mm
Resolution: 2.5 ȣm* For specification & drawing of ULE-50, please refer to page 118 for more details.

unlPULXE

www.unipulse.tokyo/en/