TM700

HIGH-SPEED TORQUE WAVEFORM MONITORFOR UTM⋜/UTMVSpecifications
Torque input

6XLWDEOHWRUTXHPHWHU

870⋜

8709

‫ ە‬High-speed sampling at 20000 times/sec.
‫ ە‬A torque curve monitor supporting thehigh response time of UTMϩ/UTMV (1 kHz).
‫ ە‬Monitoring of torque, rotation speed and power*1 curves.
‫ ە‬Upper/lower limits & hold function.
‫ ە‬Easy connection to UTMϩ/UTMV with the supplied cable.
(Operation power can be supplied to the UTMϩ/UTMVand to an optional rotary encoder*2)‫ ە‬Data recording with USB.

- Voltage input
Signal input range -5 to +5 V Input impedance: 1 MΩ or more
AccuracyNon-linearity: Within 0.02٫FS±1 digit
Zero drift: Within 0.2 mV/Υ RTI
Gain drift: Within 0.01٫/ΥAnalog filter
Low pass filter (-6 dB/oct.)
Selectable from 10 Hz, 30 Hz, 100 Hz, 300 Hz, 1 kHz, 3 kHz, 10 kHz, 30 kHz
A/D converter
Speed: 20000 times/sec.
Resolution: 24 bit (binary) Effective Resolution: Approx. 1/30000 against 5 V
Voltage monitor output
Linked to the output voltage of the torque meter
Rotation- Pulse input (open-collector)
input■ High-speed (for standard UTMϩ/UTMV)
Conforming to the maximum rotation speed of UTMϩ/UTMVMaximum input frequency
Selectable from 15, 10, 5, 3, or 2 rpm (when pulse rate is 4 ppr)
Minimum input frequency
Selectable from 60, 40, 20, 12, or 8 rpm (when pulse rate is 1 ppr)
50 ȣsMinimum detection pulse widthCircuit configuration No-voltage contact input (minus common) Open collector connectable (Ic = Approx. 10 mA)■ Low-speed‫ق‬for UTMϩwith rotary encoder option‫ك‬
Approx.1000 rpm
Maximum input frequency
0.1 rpm
Minimum input frequency
5 ȣsMinimum detection pulse widthCircuit configuration No-voltage contact input (minus common) Open collector connectable (Ic = Approx. 10 mA)
Display
Display unit
3.5 inch TFT color LCD
External
External output (8)I/O signalex) high and low limit outputs
External input (10)ex) hold control input, digital zero input (DZ)
Interface
SIF: 2-wire serial interface
USB: USB interface
D3V: D/A converter voltage output (3 ch) (Option)
Power supply
DC 24 V (±15%)Power consumption 6 W typ.
OperatingOperation temperature: ٕ10to ٔ40Υ‫ؙ‬Storage temperature: ٕ40to ٔ80Υ
conditions
Humidity: 85% RH or less (non-condensing)
Dimension96(W) × 96(H) × 138(D) mm (Not including projections)
WeightApprox. 1.0 kg
Attachments

Operation manualҩҩҩ1, External input/output connectorҩҩҩ1,Cable for connecting UTMϩ/UTMV 2 mҩҩҩ1, Operating tool (with D/A converter (3 ch) option)ҩҩҩ1

Optional
accessories

DTC2:
CA372-I/O:
CA81-USB:
CATM21-M:
CATM51-M:
CATM(R)21-M:
CATM(R)51-M:
CN50:
CN55:
CN73:
CN90:
CN91:
GMP96x96:
TSU03:

CE marking
certification

EMC directives EN61326-1*1 Power (W) = 2ȧ× Torque (N m)× Rotation speed (rpm)/ 60*2 Rotary encoder option is not available for UTMV.
It will be a waveform display of torque fluctuations in chronological order.

TorqueCase for TM700 (with AC power supply)
Cable with FCN connector at one-end 3 m
miniUSB-computer USB cable 1.8 m
Cable for connecting UTMϩ/UTMV 2 m (Same as the attachment)
Cable for connecting UTMϩ/UTMV 5 m
Cable for connecting UTMϩrotary encoder 2 m
Cable for connecting UTMϩrotary encoder 5 m
FCN series I/O connector (with cover) (Same as the attachment)
FCN series I/O connector (with diagonal cover)D/A converter (3 ch) connector
Waterproof plastic connector for UTMϩ/UTMVWaterproof plastic connector for UTMϩrotary encoder
Rubber seal
DC lighting surge unit

* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,
fluorescent display tubes and LCDs due to manufacturing process or production lots.

↓Structure of product codeTM700ձ
↑Speed↑Monitor the fluctuationof torque, rotation speed and power.

Powerձ Standard unit‫ڧ‬ղղ Interface

SignInterface
Standard SI/F, USB
D3V
D/A converter (Voltage output) (3 ch)External dimension(Front)(Side)(Rear)

Panel cutout size
Panel thickness
1.6 to 3.2 mm

Unit: mm

DTC2: Case for TM700 (with AC power supply)

120

unlPULXE

143

2m

15°

207

168
190

At stand use

www.unipulse.tokyo/en/