F372A

GRAPHIC DISPLAY / TOUCH PANEL TYPE
DIGITAL INDICATOR● 2000 times/sec. high-speed processing● Work no. can set up up to 16 types● Can start measurement even there is no external signal● Analog monitor output
Voltage output is proportionate to the input signal making
the recording on recorder convenient.
Approx. 2 V per 1 mV/V strain gauge input● A variety of interfaces
RS-232C/ CC-Link/ DeviceNet/
D/A output/ BCD output (Sink type/ Source type)unlPULXE● 3.5 inch color LCD module & touch panel‫ ؙ‬Setting operation made easy via direct touch on the touch panel.
● Multi calibration function
Stores calibration values for 4 ch portions and can be selected
via touch panel or external signal● Alarm function‫ ؙ‬Monitors if the measured value is abnormal‫ ؙ‬- Hi/Lo limit for in comparison setting - Overflow‫ ؙ‬- A/D input range- Digital zero regulation value● Can confirm forcs pattern before measurement start. (Pre Trigger)www.unipulse.tokyo/en/

Excitation voltage

DC 10 V, 2.5 V±5% (depending on settings) Output current: Within 120 mA

Signal input range−3.0 to ＋3.0 mV/VAccuracy

Non-linearity: Within 0.02% FS±1 digit (at 3.0 mV/V input)
Zero drift: Within 0.5 μV/ºC RTI
Gain drift: Within 0.01%/ ºC

Analog filter

Low pass filter (-6 dB/oct.) Selectable from 30 Hz, 100 Hz, 300 Hz, 1 kHz

A/D converter

Speed: 2000 times/sec.
Resolution: 24 bit (binary) Approx. 1/30000 at 3.0 mV/V input

Analog monitor output Output level: Approx. 2 V per 1 mV/V input ; Load resistance: 2 kΩ or more
Display

Display unit

TFT color LCD

Display area

71(W) × 53(H) mmDot structure

320 × 240 dot

Measured value

5 digits: −99999 to ＋99999 Sign: Minus sign on most significant digit

Hold1) Sample; 2) Peak; 3) Bottom; 4) P-P; 5) Average; 6) Inflection Point; 7) Relative Maximum;
8) Relative Minimum; 9) Relative Difference; 10) Sample & Peak; 11) Sample & Bottom;
12) Sample & P-P; 13) Sample & Average; 14) Sample & Inflection Point; 15) Sample & RelativeMaximum; 16) Sample & Relative Minimum; 17) Sample & Relative Difference;18) Peak & Bottom; 19) Peak & P-P; 20) Bottom & P-P; 21) Average & Peak;
22) Average & Bottom; 23) Average & P-P; 24) Relative Maximum & Relative Minimum;25) Relative Maximum & Relative Difference; 26) Relative Minimum & Relative Difference

Comparison
function

Can set 4 different settings from Hi limit, Lo limit, etc

Calibration
value
selection

Stores up to 4 types of calibration values that can be interchanged

External
signalExternal output signal (8)HI/LO comparison output (HH, HI, OK, LO,LL)/ RUN output/
Hold end output/ Graph plotting end output
Vceo = 30 V (max), Ic = 30 mA (max)External input signal (10)Work selection input/ Hold control input/ Digital zero input (DZ)/
Graph plotting control input/ Calibration selection input
Ic = 10 mA or lessInterface

Attachments FCN series I/O connector (with cover) .............……………………………………… 1
Operation manual .............………………………………………………………….…… 1
Analog I/O connector terminal block (Already mounted on the main unit) ...…… 1
BCD output connector (when BCD output option is selected) ..............…………… 1
Mini driver (when D/A converter option is selected) ..............……………………….. 1
CC-Link connector (when CC-Link option is selected) .............………………...…. 1
DeviceNet connector (when DeviceNet option is selected) ..............…………...…. 1
Optional
CA372-I/O:
Cable with FCN connector at one-end 3 maccessories CA600-BCDCNV: FCN connector 32p-57・36p cabtire cable 0.3 m
CA81-232X:
miniDIN-D-Sub9p cross cable 1.5 m
CN50:
FCN series I/O connector (with cover)(Same as the attachment)
CN51:
BCD output connector
CN55:
FCN series I/O connector (with diagonal cover)
CN60:
Round DIN 8p connector for RS-232C
CN71:
CC-Link connector
CN72:
Double row connector for CC-Link
CN80:
Analog I/O connector terminal block (Same as the attachment)
CND01:
DeviceNet connector
DTC2:
Case for F372A (with AC power supply)
GMP96x96:
Rubber packing
TSU03:DC Lightning surge unit
CE marking EMC Directive EN61326-1
certification* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,
fluorescent display tubes and LCDs due to manufacturing process or production lots.

SIF： 2-wire type serial interface
232： RS-232C communication interface
CCL： CC-Link interface (Option)
ODN： DeviceNet interface (Option)
BCO： BCD parallel data output interface (Sink type) (Option)
BSC： BCD parallel data output interface (Source type) (Option)
DAV： D/A converter voltage output (Option)
DAI： D/A converter current output (Option)*Only one option can be installedA① Standard unit

③ Interface② I/O outputSignStandard
ISCOption
ISC: I/O source board
GeneralPower supply voltage DC 24 V (±15%)specifications Power consumption 5 W typ.
Inrush current typ.
55 A, 1 msec (cold start at room temperature)
Operating conditions Operation temperature: -10 to ＋40°C
Storage temperature: -20 to ＋60°C
Humidity: 85% RH or less (non-condensing)External dimension
96(W) × 96(H) × 138(D) mm (Not including projections)
WeightApprox. 1.0 kg

DTC2: Case for F372A (with AC power supply)

Structure of product code

Output type
Sink type (NPN output)
Source type (PNP output)

SignStandardInterface
SI/F, RS-232C↓ One optional interface can be added
in addition to the standard interface.

CCL
ODN
BCO
BSC
DAV
DAI

CC-LinkDeviceNet
BCD output (Sink type)BCD output (Source type)
D/A converter (Voltage output)
D/A converter (Current output)

Unit: mm

120

1432m

15°

168
190

At stand use
207

Analog

F372A

Specifications

External dimension(Front)(Side)(Rear)

ＤＣ ＩＮ

Ｂ１

96

Panel cutout size

＋７－６５４３91

Ａ１

Panel thickness
1.6 to 3.2 mm

92 ＋1−0ＲＳ－２３２Ｃ２１Ａ１２

Ｂ１２
ＭＡＤＥ ＩＮ ＪＡＰＡＮ

96

1812091

92 ＋1−0

Unit: mmunlPULXE

www.unipulse.tokyo/en/