TM500

TORQUE WAVEFORM MONITORFOR UTMϩ/UTMVSpecifications
Input signals-5 to +5 V Input impedance: 1 MΩ or more

Torque input

AccuracyNon-linearity: Within 0.02% FS±1 digit
Zero drift: Within 0.2 mV/٦ RTI
Gain drift: Within 0.01٫/٦
Analog filter
Low pass filter (-6 dB/oct.) Selectable from 30 Hz, 100 Hz, 300 Hz, 1 kHz
A/D converter
Speed: 4000 times/sec.
Resolution: 24 bit (binary) Effective Resolution: Approx. 1/30000 against 5 V
Voltage monitor output Linked to the output voltage of the torque meter
Pulse inputMaximum input frequency: 50 kHz
Open collector) Internal counting range: Approx. 1000000
Display unit
3.5 inch TFT color LCD

Display
Setting

Setting method
Setting by analog type touch panel operation
Preservation of set values
Initial settings: NOV RAM (nonvolatile RAM),
Other settings: Lithium-battery-backed-up C-MOS RAM
External output
Hold result‫ق‬Load, DPM‫ك‬Overload / Wave result / Complete /signal (15)
RUN / Load OK / DPM OK / SD OK

External
I/O signal

External input
signal (12)6XLWDEOHWRUTXHPHWHU

870⋜8709‫ ە‬Waveform display of torque variation against angles.*‫ ە‬Suitable for low-speed rotation and linear motion applications.
‫ ە‬Saving measurement data and setting values in a SD card.
Data can be converted to CSV format and edited inExcel format easily.
‫ ە‬Easy connection to UTM⋜/UTMV with the supplied cable.

Interface

RS-232C interface

Option

SDC: SD card slot Approx. 80 waveforms can be saved by 1 MByte, SDHC and SDXC are not supported.)
All indicated values and all comparison waveforms can be saved and restored. Measuredwaveforms and judgment points can be saved automatically.

Generalperformance

Power supply
DC 24 V (±15%)Power consumption 7 W typ.

Attachments

Optional
accessories‫ ە‬The unit supplies electric power to UTM⋜/UTMV.
(Also to an optional rotary encoder*)* Rotary encoder option is not available for UTMV.
Torque fluctuation over a period of time will be displayed in a curve/ waveform.

‫ ڦ‬Application example

Reduction
gear

Motor driver
Angularposition

DTC1:
SD1G:
SD2G:
CA81-232X:
CATM21-M:
CATM51-M:
CATM(R)21-M:
CATM(R)51-M:
CN52:
CN57:
CN60:
CN90:

Torquesignal

Case for TM500 (with AC power supply)
SD card 1 GByte
SD card 2 GByte
miniDIN-D-Sub9p cross cable 1.5 m
Cable for UTM⋜UTMV connection 2 m (Same as the attachment)
Cable for UTM⋜UTMV connection 5 m
Cable for UTM⋜rotary encoder connection 2 m (Same as the attachment)
Cable for UTM⋜rotary encoder connection 5 m
FCN series I/O connector (with cover) (Same as the attachment)
FCN series I/O connector (with diagonal cover)Circular DIN 8p connector for RS-232C
Waterproof plastic connector for UTM⋜UTMV connection

Waterproof plastic connector for UTM⋜rotary encoder connection
Rubber packing
DC lighting surge unit
EMC directives EN61326-1

CE markingcertification* Please note that there are possibilities of individual differences in a color tone on display devices
such as LEDs, fluorescent display tubes and LCDs due to manufacturing process or production lots.

Structure of product code

UTM⋜
(with rotaryencorder option)

GO/NOGOsignal
TM500

Operating conditions Operation temperature: ٕ10 to ٔ40٦‫ؙ‬Storage temperature: ٕ20 to ٔ60٦
Humidity: 85% RH or less (non-condensingDimension
96(W) × 96(H) × 117.3(D) mm (Not including projections
WeightApprox. 1.0 kg
Operation manualҩҩҩ1, External input/output connectorҩҩҩ1,Cable for UTM⋜UTMV connection 2 mҩҩҩ1, Cable for UTM⋜rotary encoder connection 2 mҩҩҩ1,SD card 1 GByte (with SDC option)ҩҩҩ1

CN91:
GMP96x96:
TSU03:Motor
PLC etc.

Load Digital Zero / DPM Positioning / Start / Stop / Hold /Reset / Backlight On / Prohibit Touch Panel / Work change

TM500
ձձ Standard unit‫ٸ‬ղղ SD card slot
SignStandardSDCCard slot
W/OSD card slot (1 GByte attached)External dimension(Front)(Side)(Rear)

96

Panel cutout size

96

Panel thickness
1.6 to 3.2 mm

1899.592 +1-0

MONITOR
TM500 TORQUE

92 +1-0

DTC1: Case for TM500 (with AC power supply)

160
182

Unit: mm120

unlPULXE

1362m

www.unipulse.tokyo/en/