TM201

Suitable torque meter

UTM⋜

USB INTERFACE DEVICE FOR UTMϩ/UTMVSpecifications

UTMV‫ ە‬Variations of torque, rotation speed, and power* can bemonitored and saved on PC. (Application software canbe downloaded from our website for free)‫ ە‬Maximum, minimum, and average value can bedisplayed.
‫ ە‬Measurement (numeric) data are automatically savedin CSV format.
‫ ە‬The unit supplies electric power to UTM⋜/UTM9.
‫ ە‬Two cables are attached: one for UTM⋜/UTM9andthe other for PC.

UTM⋜/UTMV - Torque sensor input (voltage input)
Interface
Signal input range ٕ5 to ٔ5 VInput impedance: 1 MΩ or more
AccuracyNon-linearity: Within 0.02% FS±1 digit
Zero drift: Within 0.2 mV/٦ RTI
Gain drift: Within 0.01%/٦Analog filter
Primary low-pass filter 1 kHz (fixed)
Secondary low-pass filter fc = 3, 30, 300 Hz, OFF (depending on settings)
Digital filter
Data output rate
Speed: 300 times/sec.
Resolution: 24 bit (binary) Effective resolution: Approx. 1/30000 against 5 V- Pulse input for rpm (input for open collector type) Maximum input frequency Conforming to the maximum rotation speed of UTM⋜/UTMV
Minimum input frequency Selectable from 15, 10, 5, 3, or 2 rpm (when pulse rate is 4 ppr)
Selectable from 60, 40, 20, 12, or 8 rpm (when pulse rate is 1 ppr)
Minimum pulse width
50 ȣs
Circuit layout
No-voltage (dry) contact input (minus common)
open collector outputs can be connected (Ic = Approx. 10 mA)- Power supply for UTM⋜8709
Power supply
DC 24 V (UTM⋜/UTMV 1 unit)
Display
LED status light
LED (Red): power supply/alarm status
LED (Green): UTM⋜/UTMV is operating normally
InterfaceUSBGeneralPower supply
AC 100 to 240 V (ٔ10%ٕ15%  free power source 50/60 Hz)
performance* When supplied AC adapter is used
Power consumption 4 W typ. (AC adapter)
OperatingOperation temperature: 0 to ٔ40٦Storage temperature: ٕ10 to ٔ60٦
conditions
Humidity: 80% RH or less (non-condensing)
Dimension
50(W) × 80(H) × 23(D) mm (Not including projections)
WeightApprox. 120 g
Attachments Setup guideҩҩҩҩҩҩ1
Cable for connecting UTM⋜/UTMV 2 mҩҩҩҩҩҩ1AC adapter for TM201 1.8 mҩҩҩҩҩҩ1miniUSB -USB cable 1.8 mҩҩҩҩҩҩ1Optional
accessories

CA81-USB:
CATM21-M:
CATM51-M:
CN90:
TM201 AC PIN EU:

miniUSB-computer USB cable 1.8 m (Same as the attachment)
Cable for connecting UTM⋜/UTMV 2 m (Same as the attachment)
Cable for connecting UTM⋜/UTMV 5 m
Waterproof plastic connector for UTM⋜/UTMV
Switching AC PIN (EU)CE marking
certification

EMC directives EN61326-1
Safety standard EN62311
Please specify requirement for CE marking certified product when making your order.

External dimension
Unit: mm

* Power (W) = 2ȧ× Torque (N m) × Rotation speed (rpm) / 60

±5 V output
80

TM201

Supplying powerfor UTM⋜8709

UTM⋜8709

㸦87.5㸧

‫ڦ‬Application example

Power supply50

Storing measured
data in CSV format

Receiving datafrom UTM⋜8709

23.PC

Application software for USB interfaces‫ ە‬Supporting USB interface option for TM201, TM400, TM700, TM301, TM320, TM380, and TC80.
‫ ە‬Enabling real-time display of torque measured by UTMϪ/UTMϩ/UTMV, parameter settings, calibration, and data recording in
CSV format. (UTM⋝ can only use TM320, TM380 & TC80)
Software for USB interfaces for TM201‫ ڦ‬Waveform settings
Change settings of waveform display and/or data

Torque, rotation speed, power
are displayed simultaneously.

Waveform settings(1) Display mode
Normal: Waveforms currently acquired are displayed
History: Waveforms previously saved are displayed(2) Acquisition mode
Single: Once "Start" button is clicked, waveform data is acquired for pre-set time period.

Cursors

Repeat: "Single" mode operation will repeat in cycle until "stop" button is clicked.
Hold trigger (Only with TM301, TM320 and TM700): Data is acquired when "Hold" button is clicked.
Level trigger*: Pre-trigger (0 to 99), trigger data (torque, rotation speed, power),
Level (±99999), Trigger slope (positive, negative, either),
Repeat trigger (Standby & wait for another trigger after waveform acquisition)* Depending on the torque monitors, the function varies. See Help of each software.

‫ ڦ‬Cursor

Move two cursors freely, display each data value and their difference‫ ڦ‬Zoom-in display (TM201, TM301, TM320, TM400 & TC80)* PC software can be downloaded from UNIPULSE website.

unlPULXE

Zoom-in display at any area

www.unipulse.tokyo/en/