TM400

PORTABLE TORQUE MONITOR FOR UTMϩ/UTMVSpecifications
Torque input- Voltage input
Signal input range-5 to +5 V Input impedance: 1 MΩ or more

AccuracyNon-linearity: Within 0.02٫FS±1 digit
Zero drift: Within 0.2 mV/ΥRTI
Gain drift: Within 0.01٫/Υ

Analog filter

Low pass filter (6 dB/oct.)
Selectable from 10 Hz, 30 Hz, 100 Hz, 300 Hz, 1 kHz, 3 kHz, 10 kHz, 30 kHz

A/D converter

Speed: 20000 times/sec.
Resolution: 24 bit (binary) Effective Resolution: Approx. 1/30000 against 5 VRotation- Pulse input (open-collectorinput

Maximum input frequency: Conforming to the maximum rotation speed of UTM⋜/UTMV
Minimum input frequency: 15 rpm
Minimum detection pulse width: 50 ȣs
Circuit configuration No-voltage contact input (minus common)

Encoder input

Open collector connectable (Ic = Approx. 10 mA)‫ؙ‬- Pulse input (open-collector

Display

Display unit

128 × 64 dots monochromatic LCD (display area: 57 × 28 mm)

Displayed data

Torque, rotation speed, angle (numeric),

Maximum input frequency: 50 kHz

torque-time or torque-angle/displacement curve (graphic),and status display (HI, LO, OK, and HOLD)
InterfaceUSB

Power supply

Built-in secondary battery
Continuous use time: Approx. 5 hours (at room temperature)
AC adopter

OperatingOperation temperature: 10 to +40Υ

conditions

Storage temperature:
Humidity:

6XLWDEOHWRUTXHPHWHU

870⋜

8709

‫ ە‬Convenient portable monitor for easy use of UTM⋜/UTMV.
‫ ە‬Ideal for on-site calibration of nut runners(use with UTM⋜ with the square drive option).
‫ ە‬With the built-in battery, it can be operated
even without power source.
‫ ە‬Easy connection to UTM⋜/UTMV with the supplied cable.
‫ ە‬Operation power can be supplied to the UTM⋜/UTMV.
(Also to an optional rotary encoder*)88(W) × 140(H) × 35(D) mm (Not including projections

WeightApprox. 500 g

Attachments

AC adopterҩҩҩҩҩҩ1
Operation manualҩҩҩҩҩҩ1
Cable for connecting UTMϩ/UTMV 2 mҩҩҩҩҩҩ1
Cable for connecting UTMϩrotary encoder 2 mҩҩҩҩҩҩ1

Optional

CA81-USB:

accessories

miniUSB-computer USB cable 1.8 m

CATM21-M:

Cable for connecting UTMϩ/UTMV 2 m (Same as the attachment)

CATM51-M:

Cable for connecting UTMϩ/UTMV 5 mCATM(R)21:

Cable for UTMϩwith rotary encoder connection (non-connector at one end) 2 m

CATM(R)21-M:

Cable for connecting UTMϩrotary encoder 2 m (Same as the attachment)

CATM(R)51-M:

Cable for connecting UTMϩrotary encoder 5 m

CN90:

Waterproof plastic connector for UTMϩ/UTMV

CN91:

Waterproof plastic connector for UTMϩrotary encoder

TM400 AC CABLE EU: AC cable (for EU)
EMC directives EN61326-1
Safety standard EN62311
Please specify requirement for CE marking certified product when making your order.

CE markingcertification* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,
fluorescent display tubes and LCDs due to manufacturing process or production lots.

‫ ە‬High-speed sampling at 20000 times/sec.
‫ ە‬Torque, rotation speed and torque-angle curve
can be monitored.
‫ ە‬Data recording with memory function and USB.

External dimension㹎㹍㹕㹃㹐㹓㹒㹋

㹃㹌㹁㹍㹂㹃㹐

㹓㹑㹀

㹂㹁㸯㸰㹔

‫ ە‬Various display modes: real-time, graph, recorded data, etc.
‫ ە‬Upper/lower limit & hold function.

20 to +60Υ
85% RH or less (non-condensing)

Dimension‫ ە‬Change the display mode by pressing the ESC button.

Value display

㹑㸿㹔㹃㹆㹍㹊㹂㹘㹃㹐㹍

TM400 PORTABLETORQUE MONITOIR㸫㸩Ser 㸬 No㸬
MADE IN JAPAN140148

Graph display

㹁㹆㸿㹐㹅㹃

TM400 PORTABLETORQUE MONITOR㹃㹑㹁* Rotary encoder option is not available for UTMV.
It will be a waveform display of torque fluctuations in chronological order.

㹑㹃㹒883539

Unit: mm

unlPULXE

www.unipulse.tokyo/en/