Quintix® ProRedefining the Standard

Quintix® Pro
Designed to be a True
Universal Balance
Quintix® Pro Laboratory Balances deliver theperformance you need, with added premiumfeatures that raise the bar in usability, flexibility,and eco-conscience design.

Useability--

Enhanced user experience with new premiumfeatures including:Easy access to menu option with a large, 7-inch, highresolution full-touch graphic display with modern
upgraded user interface
Ensure the highest weighing accuracy with the internal
motorized levelling (optional) and automatic internal
adjustment isoCAL
Quick access to the integrated digital manual forcommon help topic
Fast adaptation of balance performance to theunique needs of each application through the
improved semi-automatic filter adaption (AFA)
Seamless connectivity features include interface
options, easy data transfer, and GLP | GMP
compliant printout

Find out more  
For more information, please visit
www.sartorius.com

Flexibility

Sustainability

Flexibility for use in a wide range of environments
and applications:Customization with different tiers of user
management access with passcode protection
Tested for use in cleanrooms grade A and B, and special
environments e.g., vacuum, argon, and nitrogen.
Smart design and chemically-resistant surfaces
made of PBT, glass and stainless steel support
routine cleaning protocols
15 integrated application programs for common
weighing tasks
Extended language package with 15 language options

Reduced environmental impact from themanufacturing plant to the bench:
Climate-conscience production process utilizes
recycled materials e.g., aluminum, in addition toenvironment-friendly packaging
Power-saving operation with stand-by or
switch-off modes
Automatic dim function, reduces display brightness
when balance is not in use to save power
User-defined, automatic start timer powers up
the instrument in time for daily use
Digital user manual minimizes paper consumption,
while allowing quick access to information------

Modern User Interface
Effortless Operation
Using the Quintix® Pro is effortless thanks to a modern
usability concept. The startup assistant allows quick and easy
setup of the balance, while the built-in interactive manual isready to assist, whenever you need help. With 15 available
language options, the Quintix® Pro is the perfect fit for anylab.

Status Center
The Status Center on Quintix® Pro provides acomprehensive summary of the balance's mostimportant status information with just a single click.

User Management
The advanced user management system in Quintix®
Pro enables you to assign individual roles to each userand customize them according to specific user needs.
This feature allows Quintix® Pro to limit menu accessstrictly to authorized personnel, thereby providing
comprehensive control and protection over yourweighing procedures.

Correct Levelling
Level Status at a GlanceWith Quintix® Pro it is easy to verify if your balance is leveled correctly.
A quick glance at the conveniently positioned level indicator is all you need.

Continuous Real-Time Level Monitoring
The LevelControl function of Quintix® Pro guarantees aperfectly leveled balance. It continuously monitors the levelstatus in real time, instantly detecting any deviations. Should
any tilt be detected, it responds immediately to safeguard
your data: it will alert you and temporarily block data transfer
to a printer or computer. Quintix® Pro will then guide you
through the leveling procedure in real time.
With the Quintix® Pro LevelControl, you can rest assured
that your measurements are completely accurate.

Motorized Levelling
Leveling is even easier with the motorized feet.
The Quintix® Pro levels itself fully automatically at thepush of a button.

Monitoring Functions and Security
isoCAL: Reliability Guaranteed
Temperature fluctuations can compromise the precision of your weighing results. Quintix® Pro eliminates this risk fromthe outset with its temperature and time-controlled calibration and adjustment function, isoCAL. By customizing the timeinterval for isoCAL, you can ensure compliance with your Standard Operating Procedures (SOPs).

Enhanced Traceability
The Quintix® Pro's internal Device-Log function
records all calibration and leveling procedures, as wellas any changes to the balance and weighing settings.
This creates a traceable log file with date and timestamps, enhancing your quality assurance processes.

Secure Weighing Data Continuously
Take advantage of the various safety level settings available on Quintix® Pro for secure data transfer. You can tailor Quintix®
Pro's automatic monitoring systems to meet your specific security requirements. Whether you choose a high-, standard-,
or low-level monitoring, Quintix® Pro ensures that only accurate data is processed.

Improved Auto Adaption toEnvironmental Conditions (AFA)
Whether your environment is stable or unstable,Quintix® Pro can be easily adapted to specificenvironmental conditions with just a single click.

Protection and Cleanability
Tested for Use in Special Environments e.g., Vacuum, Argon, and Nitrogen

Extended Protection with
In-Use-CoverEnhance the protection of Quintix® Pro by utilizing anin-use cover. This cover envelops the entire housingof the balance, offering additional defense against
contamination.
The additional protection covers for the weighing
pan and draft shield base plate is particularly usefulin areas where hazardous materials are handled.

Easy Cleaning
The contemporary design of Quintix® Pro facilitateseasy and straightforward cleaning of the balance.
The use of durable materials such as polybutylene
terephthalate (PBT), stainless steel, and glass not only
ensures high chemical resistance, but also prevents
cross-contamination.

Documentation and Data Transfer:Easy, Fast and Reliable
Be future ready! Quintix® Pro supports multiple connection to external devices via three USB type C interfaces, RS232 9-pin
interface, and an ethernet interface. Transfer weighing data or calibration reports to a USB stick or directly to a database via
the ethernet interface.

Data Transfer: Direct and Correct

Data Documentation: Fast and Easy

Effortlessly connect your Quintix® Pro to your computer orlab printer using the balance's USB Type-C interface port
and Plug & Play technology. Its PC-Direct feature allows forthe seamless transfer of your measurement results directly toyour preferred Microsoft® Office program. This eliminates theneed for additional software and prepares your data in yourchosen format, ready for calculations in spreadsheets.

When directly connected to a Sartorius laboratory printer,Quintix® Pro generates fast and accurate printouts of yourweighing results. You can use standard printer paper orpractical, continuous-length paper label rolls, or even faderesistant paper for reliable records with unique identificationof your samples.

Sample Identification: Even More Reliable and Precise with Quintix® Pro
If you're working with samples that require unique identification in regulated environments, you'll appreciate the built-in
sample identification feature of Quintix® Pro. This function allows you to tailor the printout information to your specific needs.
With an interactive touchscreen display and alphanumeric input capability, Quintix® Pro simplifies the process of entering
complex sample IDs, batch IDs, or project IDs. Generate reliable and accurate sample records with ease using Quintix® Pro.

Universal Applications
Integrated Application Programs for Efficient Working Routines
The combination of user-friendly design and diverse integrated application programs make Quintix® Proa truly universal balance that is suitable for any environment:
Weighing | Dosing, Mass unit conversion, Counting, Percentage weighing, Calculation, Mixing, Components,Density determination, Statistics, Check weighing, Animal weighing, Peak hold, Pipette smart check,
Differential weighing, Underfloor weighing.

Service Tools
Flexible Maintenance and Reliable Backup
Define service and maintenance interval for each Quintix® Pro individually to maintain compliance with your existing SOPs.
You can easily export a backup of the balance settings to external storage devices, like a USB stick. Saved records are always
accessible and can also be transferred to other devices.
Each Quintix® Pro comes with an individual factory acceptance test report, confirming compliance with all specifications.

Life Cycle Services for Your Quintix® Pro
Weighing is a critical process in all laboratories, where the reliability and reproducibility of results as well as precision andaccuracy are of the utmost importance. In order to consistently achieve trustful weighing results throughout the entire
equipment life cycle, at Sartorius high quality products go hand in hand with professional life cycle services.
We are proud to offer an extensive range of certified and accredited services for your laboratory balances, aimed at ensuring
the utmost quality of your weighing results, prolonging the equipment longevity, and minimizing downtimes. Our services
ensure that your balances operate flawlessly within their tolerance limits and obtain constantly compliant, reproducible andreliable measurement data.

Our Services at a Glance
Installation and commissioning:
Reliable results right from the start---Your advantage:
Outstanding weighing performance, steadfast results from the start
Seamless integration of equipment into your laboratory environment
Optimum system performance and results
Expertly trained operating personnel

Equipment qualification (IQ | OQ):
Compliance with GMP requirements---

Your advantage:Full compliance with regulatory requirements (GMP | GLP)
Audit security
All critical equipment aspects regimented due to our risk-based approach
Service performed with all required regulatory and technical knowledge

Preventative maintenance and Service Contracts:
Long-term process safety and reliable results---Your advantage:
Optimal operation of your system
Extended instrument lifetime
Compliance with regulatory requirements
Full cost control, with service contracts tailored to your specific needs

Calibration Services:
Traceability of weighing results, regulatory compliance---Your advantage:
Accredited calibration certificates with ISO/ IEC 17025 in most countries
Guaranteed traceability to national standards
Minimum weight certificates according USP <41> and Ph. Eur. 2.1.7
Certificates for type approved balances according to OIML R76

Technical Specifications
Analytical Balances
Model

Levelling
featureQTX324IRU-1x
QTX324IMU-1x
QTX224IRU-1x
QTX224IMU-1x
QTX124IRU-1x
QTX124IMU-1x

Weighing
capacity[g]

Readability Repeat[mg]ability[mg]

Repeatability, Stabilization
typicaltime, typical
[mg]
[s]

Weighing
pan size[mm]

Weighing
chamberheight*[mm]

Dimension(W × D × H)
[mm]

Level support
320
Motorized feet

0.1

± 0.1± 0.08

≤1.5

Ø 90

240

220 × 377 × 346

Level support
220
Motorized feet

0.1± 0.1± 0.08

≤1.5

Ø 90

240

220 × 377 × 346

Level support
120
Motorized feet

0.1

± 0.1

± 0.08

≤1.5

Ø 90

240

220 × 377 × 346

Milligram Balances
Model

Levelling
featureQTX1503IRU-1x
QTX1503IMU-1x
QTX1203IRU-1x
QTX1203IMU-1x
QTX623IRU-1x
QTX623IMU-1x
QTX423IRU-1x
QTX423IMU-1x
QTX323IRU-1x
QTX323IMU-1x

Weighing
capacity[g]Readability Repeat[mg]ability[mg]

Repeatability, Stabilization
typicaltime, typical
[mg]
[s]

Weighing
pan size[mm]

Weighing
chamberheight*[mm]

Dimension(W × D × H)
[mm]

Level support
1500
Motorized feet

1±1± 0.5

≤1

Ø 120

240

220 × 377 × 346

Level support
1200
Motorized feet

1±1± 0.5

≤1

Ø 120

240

220 × 377 × 346

Level support
620
Motorized feet

1±1± 0.5

≤1

Ø 120

240

220 × 377 × 346

Level support
420
Motorized feet

1±1± 0.5

≤1

Ø 120

240

220 × 377 × 346

Level support
320
Motorized feet

1±1± 0.5≤1Ø 120240

220 × 377 × 346* Upper edge of the weighing pan to the lower edge of the upper draft shield panel
x = Term for country-specific models

Precision Balances
Model

Levelling
featureQTX6202IRO-1x
QTX6202IMO-1x
QTX4202IRO-1x
QTX4202IMO-1x
QTX3202IRO-1x
QTX3202IMO-1x
QTX2202IRO-1x
QTX2202IMO-1x
QTX1202IRO-1x
QTX1202IMO-1x
QTX12201IRO-1x
QTX12201IMO-1x
QTX10201IRO-1x
QTX10201IMO-1x
QTX8201IRO-1x
QTX8201IMO-1x
QTX6201IRO-1x
QTX6201IMO-1x
QTX3201IRO-1x
QTX3201IMO-1x

Weighing
capacity[g]

Readability[mg]

Repeatability[mg]

Repeatability,
typical[mg]

Stabilization
time, typical[s]

Weighing
pan size[mm]

Dimension(W × D × H)
[mm]

Level support
Motorized feet

6200

10± 10±5≤0.9

182 × 182

215 × 377 × 95

Level support
Motorized feet4200

10± 10±5≤0.9

182 × 182

215 × 377 × 95

Level support
Motorized feet3200

10± 10±5≤0.9

182 × 182

215 × 377 × 95

Level support
Motorized feet

2200

10± 10±5≤0.9

182 × 182

215 × 377 × 95

Level support
Motorized feet

1200

10± 10±5≤0.9182 × 182

215 × 377 × 95

Level support
Motorized feet

12200

100± 100± 50≤0.9182 × 182

215 × 377 × 95

Level support
Motorized feet

10200

100± 100± 50≤0.9182 × 182

215 × 377 × 95

Level support
Motorized feet

8200

100± 100± 50≤0.9182 × 182

215 × 377 × 95

Level support
Motorized feet

6200

100± 50± 50≤0.9182 × 182

215 × 377 × 95

Level support
Motorized feet

3200

100± 50± 50≤0.9182 × 182

215 × 377 × 95

* Upper edge of the weighing pan to the lower edge of the upper draft shield panel
x = Term for country-specific models

Germany
Sartorius Lab Instruments
GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0
For further information, visit	
www.sartorius.com

Specifications subject to change without notice.
©2023 Sartorius Lab Instruments GmbH & Co. KG.
Status: 12 | 2023

USA
Sartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 368 7178