Product Datasheet

Quintix® Pro--Benefits
Enhance the user experience with new premium features
Offering a wide range of use in different environments
and applications
Energy efficiency, reduced carbon footprint, andsustainability, minimizing environmental impact fromthe manufacturing plant to the bench

Product Information
It’s easy to overlook the standard lab balance. To expect theordinary. Until something new comes along that redefines
what a standard balance can be.
Quintix® Pro Laboratory Balances deliver the performance
you need, with added premium features that raise the bar inusability, flexibility, and eco-conscience design.

Technical Specifications
General Technical Data
Ambient Conditions

Value

Installation site

For indoor use only, max. height above sea level3000 mTemperature

Environment (metrological data)*
Environment
Storage and transport+10 – +30° C+5 – +40° C-20 – +60° C

Relative humidity**

At temperatures up to 31° C, non-condensing, then linear decrease
from max. 80% at 31° C to max. 50% at 40° C

80%

No heat from heating systems or direct sunlight
No electromagnetic fields
Power Supply Device

Value

Input voltage

15 VDC (±10%)Power consumption, max

4W

Only by Sartorius power supply unit YEPS01-15V0W with interchangeable country-specific plug-in AC adaptors
Power Supply Unit

Value

Type: Sartorius power supply unit YEPS01-15V0W
PrimaryVoltage
Frequency
Current consumption, maximum

100 – 240 VAC (±10%)
50 - 60 Hz
0.2 A

SecondaryVoltage
Current, maximum

15 VDC (±5%)
0.53 A

Short-circuit protection

Electronic

Protection class according to IEC 60950-1

IIPollution level according to IEC 61010-1

2

Overvoltage category according to IEC 60664-1

II

Other data: See label on the power supply unitElectromagnetic Compatibility
Interference resistance: Suitable for use in industrial areas
Transient emissions

Class B
Suitable for use in residential areas and areas that are connected toa low voltage network that also supplies residential buildings

Materials
Housing

Polybutylene terephthalate (PBT)

Control module

Glass

Draft shield

Glass | polybutylene terephthalate (PBT)

Weighing pan

Stainless steel

Warm-up Time

Value

Device, approx

2 hr* For conformity-assessed (verified) balances in accordance with EU requirements, refer to the information on the balance.
** For conformity-assessed (verified) balances in accordance with EU requirements, the legal regulations apply.

2Interfaces
Specifications of the RS232 Interface
Type of interface

Serial interface

Interface operation

Full duplex

Level

RS232Connection

D-sub connector, 9-pin

Maximum cable length

10 m

Pin assignment

Pin 1: Not assigned
Pin 2: Data output (T × D)
Pin 3: Data input (R × D)
Pin 4: Not assigned
Pin 5: Internal groundPin 6: Not assigned
Pin 7: Clear to Send (CTS)
Pin 8: Request to Send (RTS)
Pin 9: Not assigned

Specifications for the USB-C Interface (back)Communication

USB UTL

Connectable devices

Sartorius printers, Sartorius second display, FTDI cable or USB memory stickSpecifications for the PC-USB Interface (back)Communication

USB Device

Connectable devices

PC

Specifications for the USB-C Interface (front)--

Communication

USB UTL

Connectable devices

Sartorius printers, Sartorius second display, FTDI cable or USB memory stick

Calibration
Internal calibration isoCAL
Internal calibration

Selectable Weight Units*--

3-point calibration
External calibration

Gram, kilogram, carat, pound, ounce, troy ounce, Hong Kong tael, Singapore tael, Taiwan tael, grain, pennyweights,
milligram, parts per pound, China tael, mommes, Austrian carat, tola, baht, mesghal and Newton
Display-----

Intuitive graphic touch technology
Built-in Applications
Weighing | Dosing
Mass unit conversion
Counting
Percentage weighing
Calculation

Languages------

MixingComponents
Density determination
Statistics
Checkweighing----

Animal weighing
Peakhold
Pipette smart check
Differential weighing
Underfloor weighing

Chinese, Czech, Dutch, English, French, German, Hungarian, Italian, Japanese, Korean, Polish, Portuguese,Russian, Spanish, Turkish
Protection
Chemical resistant housing parts
Glass parts of the draft shield are coated toreduce electrostatic influences
Display foil (available as assessory)

Anti-theft Lock

Dust cover for balances (also available as accessory)
In-use-cover for housing (also available as accessory)
In-use-cover for weighing pan (available as accessory)

Kensington lock and lockdown capability for cable or chain* The availability of units depends on national legislation and is therefore country-specific.

Quintix® Pro3

Model specific data
For models with internal adjustment feature
Model QTX

Unit

324Ixy-1z¹

224Ixy-1z¹124Ixy-1z¹

Readability | Scale interval (d)

mg

0.1

0.1

0.1

Maximum capacity (Max)

g320

220

120EMC

EMC

EMC

Weighing system
Repeatability
At 5% load, typical value

± mg

0.08

0.08

0.08

At approx. maximum load, typical value

± mg

0.1

0.1

0.1

Limits

± mg

0.2

0.2

0.2

Typical value

± mg

0.06

0.06

0.06

Linearity deviation

Eccentricity deviation (Deviation when load is off-center, positions according to OIML R76)
Test weightg200

10050Tolerance

± mg

0.4

0.40.4

Typical value± mg

0.20.120.1± ppm/K

111

Sensitivity drift between +10° C and +30° C
Tare maximum capacity (subtractive)<100% of maximum capacity

isoCAL:Temperature change

K

1.5

1.5

1.5

Time interval

h444

Accuracy classIII

TypeBC-QA

BC-QB

BC-QB

For models with approval:

Verification scale interval (e)mg111Minimum load (Min)mg

101010Minimum initial weighing according to USP (United States Pharmacopeia), Chap. 41
Optimum minimum initial weighing

g0.082

0.082

0.082

Typical minimum initial weighing

g0.16

0.160.16

Typical measurement times

≤2.0

≤2.0

≤2.0

Typical stabilization times

≤1.5

≤1.5

≤1.5

g

200

200

100

E2E2

E2

Recommended calibration weight
External calibrated test weight
Accuracy class in accordance with OIML R111-1
Weighing pan size

mm

Ø 90

Ø 90

Ø 90

Weighing chamber height*

mm

240

240

240

Dimension (W x D x H)

mm

220 × 377 × 346

220 × 377 × 346

220 × 377 × 346

Net weight, approx.

kg

6.70

6.70

6.70

Gross weight, approx.

kg

8.65

8.65

8.65

* upper edge of the weighing pan to the lower edge of the upper draft shield panel Levelling, x =x = R : Real-time level support
x = M : Motorized, automatic level feet1Draft shield, y =y = U : Manual, glass
y = O : No draft shield

14¹Country-specific marking in model, z =z = S : Standard balances without country-specific additionsz = SAR : Standard balances with country-specific additions for Argentina
z = SJP : Standard balances with country-specific additions for Japanz = SKR : Standard balances with country-specific additions for South Korea
z = CEU : Conformity-assessed balances with EU type examination
certificate without country-specific additionsz = CFR : Conformity-assessed balances with EU type examination
certificate only for Francez = NUS : Balances with approval for Canada and USA
z = OBR : Balances with approval for Brazil
z = OCN : Balances with approval for Chinaz = OIN : Balances with approval for Indiaz = OJP : Balances with approval for Japanz = ORU : Balances with approval for Russia

1503Ixy-1z¹

1203Ixy-1z¹

623Ixy-1z¹

423Ixy-1z¹

323Ixy-1z¹

111111.500

1,200620

420320EMC

EMC

EMC

EMC

EMC

0.5

0.5

0.5

0.5

0.5

11111222220.6

0.6

0.6

0.6

0.6

500

500

200200200

2222211111

1,5

1,5

222

<100% of maximum capacity
1.5

1.5

2224

4666II

IIII

II

BC-QC

BC-QC

BC-QD

BC-QD

BC-QD

10101010

10

100

10020

20

20

0.82

0.82

0.82

0.82

0.82

11111

≤1.5

≤1.5

≤1.5

≤1.5

≤1.5

≤1.0

≤1.0

≤1.0

≤1.0

≤1.0

1,000

1,000

500

200

200E2E2F1F1F1

Ø 120

Ø 120

Ø 120

Ø 120

Ø 120

240

240

240

240

240

220 × 377 × 346

220 × 377 × 346

220 × 377 × 346

220 × 377 × 346

220 × 377 × 346

8.10

8.10

6.70

6.70

6.70

10.03

10.03

8.65

8.65

8.65

Quintix® Pro5

Model QTX

Unit

6202Ixy-1z¹

4202Ixy-1z¹

3202Ixy-1z¹

2202Ixy-1z¹

Readability | Scale interval (d)mg

10101010

Maximum capacity (Max)

g

6,200

4,200

3,200

2,200

EMC

EMC

EMC

EMC

Weighing system
Repeatability
At 5% load, typical value± mg

5555At approx. maximum load, typical value± mg

10101010Limits± mg

20202020

Typical value

± mg

6666

Linearity deviation

Eccentricity deviation (Deviation when load is off-center, positions according to OIML R76)
Test weight

g

2,000

2,000

2,000

1,000

Tolerance± mg

20202020

Typical value± mg

10101010± ppm/K

2222Sensitivity drift between +10° C and +30° C
Tare maximum capacity (subtractive)<100% of maximum capacity

isoCAL:Temperature change

K

2222Time intervalh6666

Accuracy classIIIIIIIITypeBC-QE

BC-QE

BC-QE

BC-QE

For models with approval:

Verification scale interval (e)

mg

100

100

100100

Minimum load (Min)

mg

500

500

500

500

Minimum initial weighing according to USP (United States Pharmacopeia), Chap. 41
Optimum minimum initial weighing

g

8.2

8.2

8.2

8.2

Typical minimum initial weighing

g

10

10

1010

Typical measurement times

≤1.0

≤1.0

≤1.0

≤1.0

Typical stabilization times

≤0.9

≤0.9

≤0.9

≤0.9

g

5,000

2,000

2,000

2,000

F1F1F1F1

Recommended calibration weight
External calibrated test weight
Accuracy class in accordance with OIML R111-1
Weighing pan sizemm

182 × 182

182 × 182

182 × 182

182 × 182

Weighing chamber height*

mm

----

Dimension (W x D x H)

mm

215 × 377 × 95

215 × 377 × 95

215 × 377 × 95

215 × 377 × 95

Net weight, approx.

kg

6.20

6.20

6.20

6.20

Gross weight, approx.

kg

8.40

8.40

8.40

8.40

* upper edge of the weighing pan to the lower edge of the upper draft shield panel Levelling, x =x = R : Real-time level support
x = M : Motorized, automatic level feet1Draft shield, y =y = U : Manual, glass
y = O : No draft shield

16¹Country-specific marking in model, z =z = S : Standard balances without country-specific additionsz = SAR : Standard balances with country-specific additions for Argentina
z = SJP : Standard balances with country-specific additions for Japanz = SKR : Standard balances with country-specific additions for South Korea
z = CEU : Conformity-assessed balances with EU type examination
certificate without country-specific additionsz = CFR : Conformity-assessed balances with EU type examination
certificate only for Francez = NUS : Balances with approval for Canada and USA
z = OBR : Balances with approval for Brazil
z = OCN : Balances with approval for Chinaz = OIN : Balances with approval for Indiaz = OJP : Balances with approval for Japanz = ORU : Balances with approval for Russia

1202Ixy-1z¹

12201Ixy-1z¹

10201Ixy-1z¹

8201Ixy-1z¹

6201Ixy-1z¹

3201Ixy-1z¹

10

100

100

100

100

100

1,200

12,200

10,200

8,200

6,200

3,200

EMC

EMC

EMC

EMC

EMC

EMC

55050

50

50

50

10100

100

1005050

20

100

100

100

100

100

6606060

60

60

500

5,000

5,000

5,000

2,000

2,000

20200

200

200

200

200

10100

100

100

100

100

244422<100% of maximum capacity
222222666666II

IIIIIIII

II

BC-QE

BC-QG

BC-QG

BC-QG

BC-QG

BC-QE

100

1,000

1,000

1,000

100

100

500

5,000

5,000

5,000

5,000

5,000

8.2

82

82

82

82

82

10

100

100

100

100

100

≤1.0

≤1.0

≤1.0

≤1.0

≤1.0

≤1.0

≤0.9

≤0.9

≤0.9

≤0.9

≤0.9

≤0.9

1,000

10,00010,000

5,000

5,000

2,000

F1F2F2F2F2F2182 × 182

182 × 182

182 × 182

182 × 182

182 × 182

182 × 182------

215 × 377 × 95

215 × 377 × 95

215 × 377 × 95

215 × 377 × 95

215 × 377 × 95

215 × 377 × 95

6.20

6.20

6.20

6.20

6.20

6.20

8.40

8.40

8.40

8.40

8.40

8.40

Quintix® Pro7

Technical Drawings
Models with a readability of 0.1 mg
All dimensions are given in millimeters

243

152

63

7

84A

991

Models with a readability of 1 mg
All dimensions are given in millimeters

AAC

126

BB

346 ± 2240B346 ± 2240

126C435 ± 2

E

150 ± 290377 ± 2

DGültig für / valid für:Bauart / TypeE

BC-QA
Material
number
1000121452 BC-QB Material number
Material number
1000121312
120

QTX324IRU-1SKR

General tolerances

Change no.

639211.

ISO 13715

83401

E 5-6.

Index

Text Material number hinzu / added

Field

Material

Changes

Weight approx.:

1-2023-06-29 BAUER-C

Part name2023-09-26 WENZEL-H

Masszeichnung__224g/324g

2023-09-26 WENZEL-H

Edges according to2.

WENZELF

1E.

Design responsible
DocumentdrawnDocument
checkedDocument
releasedTechnical specificationFD

220 ± 2D+2

41 0

234 ± 2150 ± 2377 ± 2220 ± 2+2

41 0

234 ± 2

The reproduction, distribution and utilization of this document as well as thecommunication of its contents to others without explicit authorization is prohibited.
Offenders will be held liable for the payment of damages. All rights reserved in theevent of the grant of a patent, utility model or design.

C+2Scale.
.

dimensinal drawing_224g/324g
Drawing no.

VersionSheet3151828

01

1Part doc. no.

3901

4Size1S.

A
3Models with a readability of ≥ 10 mg
All dimensions are given in millimeters

1243

5678A

95 ± 2

B+238 0237 ± 2377 ± 2

D182 ± 1
215 ± 2

The reproduction, distribution and utilization of this document as well as thecommunication of its contents to others without explicit authorization is prohibited.
Offenders will be held liable for the payment of damages. All rights reserved in theevent of the grant of a patent, utility model or design.

CGültig für / valid für:Bauart / TypeBC-QE Material number 1000121472
BC-QG Material number 1000121373

E

Material number

QTX8201IRO-1SKR

General tolerances

182 ± 1

Change no.

639211.

.

.

01

E 5-6

Text Material number hinzu / added

Index

Changes

Field

Material

Weight approx.:

Design responsible

WENZEL
Technical specification

F.

ISO 13715

134-2023-06-29 BAUER-C

Part name2023-09-26 WENZEL-H

Masszeichnung_6202g/12201g

2023-09-26 WENZEL-H

Scale

dimensinal drawing_6202g/12201g+.
.

Version Sheet

Drawing no.

Edges according to2

Documentdrawn
Document
checkedDocument
released3154178
Part doc. no.

01901

Quintix® Pro

1Size

9

Accessories
These tables contain an excerpt of the accessories that can be ordered. For information on other products, contact Sartorius.

Balance Accessories
Item

Quantity

Order number

Display protection film (set of 5)

1

YDC40

Dust cover for balances with an analytical draft shield

16960SE06

Dust cover for balances with rectangular weighing pan

16960SE07

In-use cover for balances with an analytical draft shield (1 piece)16960SE08

In-use cover for balances with an analytical draft shield (set of 5)16960SE08-5

In-use cover for balances with rectangular weighing pan (1 piece)16960SE09

In-use cover for balances with rectangular weighing pan (set of 5)

16960SE09-5

Set protection cover for 90 mm weighing pan and draft shield base plate (set of 5)

1

YIC02-A

Set protection cover for 120 mm weighing pan and draft shield base plate (set of 5)

1

YIC02-M

Set protection cover for 182 × 182 mm weighing pan (set of 5)

1

YIC02-P

Round glass draft shield for balances with a readability of 1 mg

1

YDS02QTX

for solids and liquids for balances with a readability of 0.1 mg | 1 mg

1

YDK03

for solids and liquids for balances with a readability of ≥10 mg

1

YDK04QTX

Demo | transportation case

1

YDB01QTX“Kensington Lock” anti-theft device

1YKL01

Pedal button | foot switch

1

YFS04

Second display | remote display

1

YSD01

Converter cable 5 VDC > 15 VDC

1

YCC-5V-15V

Made from wood with natural stone

1

YWT09

Made from natural stone, with vibration dampening

1

YWT03

1YWT04

Density determination set

Weighing table

Wall console made from natural stone10Printer and Accessories for Data Communication
ItemQuantity

Order number

Thermal transfer printer (USB-B, RS232

1YDP30

Thermal printer (USB-B)

1YDP40Dot matrix printer (USB-B, RS232)

1YDP50Dot matrix printer (RS232)*

1

YDP20-0CE

Data cable USB-C > USB-B (>YDP30, YDP40, YDP50), 1.5 m

1

YCC-USB-C-B

Data cable USB-C > USB-A (>PC), 1.5 m

1

YCC-USB-C-A

Data cable RS232 (9-pin) > USB-A (>PC), 1.5 m

1YCC-D09M-USB-A

Data cable RS232 (9-pin) male > RS232 (9-pin) male (>YDP30), 1.5 m

1YCC-D09MMData cable RS232 (9-pin) male > RS232 (9-pin) female (>YDP20-0CE, YSD01), 1.5 m

1YCC-D09MF

Y-adapter RS232 (9-pin) male > 2× RS232 (9-pin) female, 0.15 m

1YCC-D09M-2D09F*additional power supply e.g. YEPS01-PS4 or YEPS01-PS5 is required

External Calibration and Adjustment Weights
QTX modelWeight

Accuracy class

Order number

324 | 224

200 g

E2

YCW522-AC-02

124

100 g

E2

YCW512-AC-02

1503 | 1203

1,000 gE2YCW612-AC-02

623500 g

F1YCW553-AC-02

423 | 323

200 g

F1YCW523-AC-02

6202

5,000 gF1YCW653-AC-02

4202 | 3202 | 2202

2,000 gF1YCW623-AC-02

1202

1,000 gF1YCW613-AC-02

12201 |10201

10,000 g

F2

YCW714-AC-02

8201 | 6201

5,000 gF2

YCW654-AC-02

3201

2,000 gF2

YCW624-AC-02

2201

2,000 gF2

YCW624-AC-02

Quintix® Pro11

GermanySartorius Lab Instruments
GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0
For further information, visit	
www.sartorius.com

Specifications subject to change without notice.
©2023 Sartorius Lab Instruments GmbH & Co. KG.
Status: 12 | 2023

USASartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 368 7178