ATEX Compliant

Digital Pressure Switch for Air

Series 56-ISE70
II 3G Ex ec IIC T5 Gc 0°C ≤ Ta ≤ 50°C
II 3D Ex tC IIIC T53°C Dc IP67

How to Order1MPa

02

56 ISE70

ATEX category 3M

43

Piping specifications

Option 202
N02
F02

Nil

Rc1/4
NPT1/4

None
With bracket

G1/4 (ISO1179)

Output specifications
272 settings
NPN open collector 2 outputs (Pin no.: 2, 4)

43

Fixed setting: (The pressure set point for
switching the output signal is common toNPN and PNP.)NPN open collector 1 output (Pin no.: 4) +PNP open collector 1 output (Pin no.: 2)

65

Fixed setting
PNP open collector 1 output (Pin no.: 4)67

2 settings
PNP open collector 2 outputs (Pin no.: 2, 4)

ANote) Mounting screws
are not included.

Option 1
NilNoneS

Lead wire with M12 connector (5 m),
straightL

Lead wire with M12 connector (5 m),right-angled

Unit specifications
NilM
P

With display unit switching function Note 1)
Fixed SI unit Note 2)
Pressure unit: psi (Initial value)
With display unit switching function Note 1)

Note 1) Under the New Measurement Law, sales ofswitches with the unit switching function are notallowed for use in Japan. (Initial value: MPa)
Note 2) Fixed unit: Mpa

Specifications
56-ISE70

Model
Rated pressure range

0 to 1 MPa

Pressure display range/Set pressure range−0.1 to 1 MPa

Withstand pressure

1.5 MPa

Pressure display resolution/Minimum unit setting

0.01 MPa

Applicable fluid

Air, Non-corrosive gas, Non-flammable gas

Power supply voltage

12 to 24 VDC±10%, Ripple (p-p) 10% or less (with power supply polarity protection)

Current consumption

55 mA or less (at no load)

Follow the instructions given below when handling the pressure switch.
• Operating temperature range is 0 to 50°C
• Do not expose the pressure switch to heat radiation from a heat source located nearby. It can cause malfunction.
• Do not expose the pressure switch/connector/cable to vibration and impact. Otherwise it can cause damage or malfunction.
• Protect the product from direct sunlight or UV light using a suitable protective cover.
• Do not disconnect the M12 connector while energized.
• Use only an ATEX approved M12 connector.
• For cleaning this product, use a clean and damp cloth, to prevent the buildup of an electrostatic charge.
• Ground properly to prevent the buildup of an electrostatic charge.

All other specifications are the same as the standard products Series ISE70.
Click here for details.

97 C

ATEX Compliant

Digital Pressure Switch for General Fluids

Series 56-ISE75/75H
II 3G Ex ec IIC T4 Gc -5°C ≤ Ta ≤ 50°C
II 3D Ex tC IIIC T54°C Dc IP67

How to Order

10MPa

56 ISE75

0243

M15MPa56 ISE75H0243

M

ATEX category 3Piping specifications

Option 202
N02
F02

NilRc1/4
NPT1/4None
With bracket

G1/4 (ISO1179)

Output specifications
272 settings
NPN open collector 2 outputs (Pin no.: 2, 4)

43

Fixed setting: (The pressure set point for
switching the output signal is common to
NPN and PNP.)NPN open collector 1 output (Pin no.: 4) +PNP open collector 1 output (Pin no.: 2)

65

Fixed setting
PNP open collector 1 output (Pin no.: 4)67

2 settings
PNP open collector 2 outputs (Pin no.: 2, 4)

ANote) Mounting screws
are not included.

Option 1

Unit specifications
NilMP

With display unit switching function Note 1)
Fixed SI unit Note 2)

NilNoneS

Lead wire with M12 connector (5 m),
straightL

Lead wire with M12 connector (5 m),right-angled

Pressure unit: psi (Initial value)
With display unit switching function Note 1)

Note 1) Under the New Measurement Law, sales ofswitches with the unit switching function are notallowed for use in Japan. (Initial value: MPa)
Note 2) Fixed unit: Mpa

Specifications
Model

56-ISE75

56-ISE75H

Rated pressure range

0 to 10 MPa

0 to 15 MPa

Pressure display range/Set pressure range

0.4 to 10 MPa

0.5 to 15 MPa

Withstand pressure

30 MPaPressure display resolution/Minimum unit setting
Applicable fluid

45 MPa0.1 MPa

Fluid or gas that will not corrode SUS304, SUS430 and SUS630

Power supply voltage

12 to 24 VDC±10%, Ripple (p-p) 10% or less (with power supply polarity protection)

Current consumption

55 mA or less (at no load)

Follow the instructions given below when handling the pressure switch.
• Operating temperature range is − 5 to 50°C
• Do not expose the pressure switch to heat radiation from a heat source located nearby. It can cause malfunction.
• Do not expose the pressure switch/connector/cable to vibration and impact. Otherwise it can cause damage or malfunction.
• Protect the product from direct sunlight or UV light using a suitable protective cover.
• Do not disconnect the M12 connector while energized.
• Use only an ATEX approved M12 connector.
• For cleaning this product, use a clean and damp cloth, to prevent the buildup of an electrostatic charge.
• Ground properly to prevent the buildup of an electrostatic charge.

All other specifications are the same as the standard products Series ISE75/ISE75H.
Click here for details.

98 C