NOVEMBER

2023

OR T HOPEDIC INS T RUMEN T S®

Beicker
Hammerhead
RongeurBack Cover

Wells Modified Lambotte PAO Osteotomes
Page 14

Featuring
many

instruments
throughout

Depth Gauge
Page 11

Angled Capsule
Scissors

Large T-Handle
Fixed Drivers

Page 33

Page 46

Hip Instruments
1.800.548.2362 • INNOMED.NET

Set Instruments
Socket Clamp Screw

Prototype
Shown

SocketBase Clamp

Cross Level

Offset StopFor use with lateral femoral positioned
patients in both the direct lateral andposterior hip approaches, the device isdesigned to help with intraoperative leg
length and femoral offset assessment, andcan be placed prior to dislocation of the hipand replaced following trial implantation
and reduction, and again at the time of
final implantation and reduction

Parsley Intraoperative
Leg Length/Offset Device

Post

Designed by Brian S. Parsley, MD

Outrigger

Leg Length Device
Assembly
Included in Sets

Sterilizable Levels
Included in Sets

PRODUCT NO’S:

2615-00 [Set with Case]
2615-05 [Set with Case and #8248 Fixed Driver]

Sterilization Case

Sets Include:

Included in Sets

2615-10 [Leg Length Device Assembly]
1180 [Sterilizable Level]
Two included in set; one with this product number

1015 [Sterilization Case]

Fixed Driver with
Zimmer Hall Quick-connect

Optional Items (included with Set #2615-05):

8248 [Fixed Driver with Zimmer Hall Quick-connect]

Included in Set #2615-05

Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

Surgical technique available on our website.

Figure 1

Position the patient in
the lateral decubitous
position, with thepatients hips flexed
at 35-45° and theknees at 90° flexion.
The down leg secured
with tape, and thefoot palpable throughthe drapes.

Figure 2

Figure 3

Figure 4

Threaded Post into fixed position
into ilium.

Offset Stop and Outrigger are slid
onto Threaded Post and adjusted
to mark a fixed reference point on
femur/greater trochanter.

Socket Base/Cross Level unit isinserted onto top of Threaded Post
using the reference alignment
arrows on the items.

Figure 5

Figure 6

Figure 7

If necessary, Cross Level isinserted into Socket Base.

Magnet Levels may be used on top of
Cross Level and locked into desired
position. After confirming position
and GT mark, the Socket Base can beremoved for the surgery MAINTAINING
locked position.

After implant trials are inserted,
the Socket Base/Cross Level canbe re-applied to Threaded Post andreference mark checked to determine
if length adjustments are necessary.

The standard
surgical approach is
performed.
Just prior todislocation of the hip,the legs are aligned,
and the heels are
positioned together
for consistency.

2

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Pen drawn landmark
on skin surface.
With heavier patients,
place pin at crestof pelvis or iliac crest.

Skin Incision

Wixson Leg Length Gauge
Designed by R.L. Wixson, MD

Used for interoperative leg length measurement
during minimally invasive total hip arthroplasty

Small Drill Hole In Femur

Fits in 5/64 (2 mm) drill hole in trochanter underneath fascia
and skin incision. Measures to a skin mark over the iliac
crest with the leg supported in a standardized position(e.g. resting on a Mayo stand).
PRODUCT NO’S:

Optional Pin
In Iliac Crest

1210-02 [2"]Depth: 2" (5,1 cm)
Overall Length: 8" (20,3 cm)
Length-to-bend: 7" (17,8 cm)
Pin Length: 10 mm

Anterior Hip Referencing Rod Assembly

1210-03 [2.75"]

Depth: 2.75" (7 cm)
Overall Length: 8" (20,3 cm)
Length-to-bend: 7" (17,8 cm)
Pin Length: 10 mm

Designed by Scott A. Foster, MD

For use during intraoperative imaging while performing anterior hip arthroplasty to help determine implantfit, position, alignment and recreation of leg length and offset using the contralateral hip for reference

Designed to be overlayed on the pelvis during theimaging part of the procedure to compare leg lengthand offset to the contra lateral hip using the transteardrop or trans ischial line as reference
Extended length allows the surgeons hands to remain
outside of the imaging beam
Notched in increments of 1 cm for ease of reference
Features a threaded coupler midshaft to break down
for processing and storage, allowing the unit to fit
into a traditional tray

PRODUCT NO’S:

2674-00 [Complete Assembly]
Overall Length: 27.75" (70,5 cm)
Rod Diameter: .25" (6,3 mm)

2674-A [Top Assembly]

Overall Length: 16.75" (42,6 cm)
Rod Diameter: .25" (6,3 mm)

Midshaft coupler
allows the unit to fit intoa traditional tray

2674-B [Bottom Assembly]
Overall Length: 10.5" (26,7 cm)
Rod Diameter: .25" (6,3 mm)

Mongold Capsule Knife
Designed by Evie Mongold, MD

Designed to reach behindthe femoral head to release
the capsule ligament

PRODUCT NO:

4115

Overall Length: 7.75" (19,7 cm)
Blade Diameter: 2" (5,1 cm)
Blade Width: .5" (1,3 cm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

3

Koonin Leg Length Caliper

Designed by Michael Koonin, MD

Designed to help measure and evaluate pre- andpost-THR leg length in conjunction with X-ray
calibration and clinical judgement
Utilizes a 5/32" (4 mm) pin in the wound just proximal to the acetabulum anda 1/8" (3.2 mm) pin in the greater trochanter. (The soft tissue is cleared away
and a single drill hole is made in the trochanter to accommodate the distal
pin; the hole is marked with methyline blue so it can be easily found.)
Alternatively, a 7.3 mm cannulated screw that accepts a 3.2 mm pin may
be used in the greater trochanter. Using the sliding caliper, the difference in leg
length measurement before hip dislocation and after the THR procedure helps
show the change in leg length.
The sterilizable level helps to ensure the leg is in the same plane when
initially putting the leg length caliper on and when reattaching the caliper.

PRODUCT NO’S:

1195 [Complete Set]   Includes: Caliper, Sterilizable Level, and Sterilization Case
Individual/Replacement Parts:

1195-01 [Caliper Only]

Overall Length: 4.5"-6.5" (11,4 cm-16,5 cm)

Hole for a 5/32" (4 mm)
pin in the iliac crest

1180 [Sterilizable Level Only]

Locking screw for caliper

Dimensions: 2" x .5" x .75" (5,1 cm x 1,3 cm x 1,9 cm)

1025 [Sterilization Case]

Hole for a 1/8"
(3.2 mm) pin in thegreater trochanter

Locking screw usedon screw pin at 1st
measurement toensure level with
2nd measurement

Sterilizable level
included in set. Can
be steam sterilized
without vacuum.

Koonin Leg Length Caliper – Small
Designed by Michael Koonin, MD

Designed for use in small incisions to
help measure and evaluate pre- andpost-THR leg length in conjunction with
X-ray calibration and clinical judgement
Works in a similar manner to the Leg Length
Caliper above without the use of the level.
PRODUCT NO:

1196   Overall Length: 3.25–4.5" (8,3 cm–11,4 cm)

Cannestra Hip Length Gauge
Designed by Vince Cannestra, MD

Helps determine leg lengthand hip offset in total hip
arthroplasty, including
minimally invasive techniques
A detailed instruction brochureis available on our website.
PRODUCT NO’S:

Ruler #1327-03

100 mm Pin #1327-01

T-Handle #1327-02

1327-00 [Set]
Replacement Parts:

1327-01 [Pin – 100 mm]
1327-02 [T-Handle]

Dimensions: 8" x 5" (20,3 cm x 12,7 cm)

Set consists of one Ruler, one Pin Inserter/Extractor Handle, one 100 mm Pin,one 130 mm Pin (not shown), and a sterilization case (not shown).

1327-03 [Ruler]1327-04 [Pin – 130 mm]
1025 [Sterilization Case]

4

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Llinas Leg Length & Lateral Offset Gauge

Designed by Adolfo Llinás, MD

Designed to help equalize the pre- and post-operative leg length/lateral hip offset
Used intra-operatively to establish measurements of both leg length and lateral hip offset. The measurements
can then be used for verification, after femoral stem and head implantation but before final fixation, to help
determine what adjustments (if any) are necessary.
PRODUCT NO’S:

Level

1133-00 [Set]
Set Includes:1133-01 [Gauge Body, Pin, and Tube Unit]

The sterilizable level can be steam
sterilized without vacuum for use in surgery.

1180 [Sterilizable Level]

Usage guide available at:www.innomed.net/instructions_innomed.htm

Slider Bar Length: 5" (12,7 cm)
Cannulated Tube Length: 3.95" (10 cm)

Dimensions: 2” x .5” x .75” (5,1 cm x 1,3 cm x 1,9 cm)

1025 [Sterilizable Case]

3.5 mm Pin
Partially Cannulated Tube

NEW USAGE VIDEOS
available on our website
www.innomed.net/
hip_tools_measuring.htm

Llinas Vertical Offset Gauge

Sliding
Bevel
Designed by Adolfo Llinás, MD

Designed to help equalize the pre- and post-operative vertical hip offset

Used intra-operatively to help determine the verticaldistance of offset (if any) between the rotational center ofthe femoral head and the top of the greater trochanter. The
measurement can then be used for verification, after femoral
stem and head implantation but before final fixation, tohelp determine what adjustments (if any) are necessary toequalize the pre- and post-operative rotational centertrochanteric offset.
Usage guide available at: www.innomed.net/instructions_innomed.htm“T”

PRODUCT NO:

1133-02 [Llinas Vertical Offset Gauge]

Overall Length: 17.25" (43,8 cm)
Sliding Bevel Arm Lengths: 2.4" (6 cm) / 3.15" (8 cm)

Flexible Ball Nose Reamer

Designed for safe and effective use in removing pedestal
formation in the femoral and tibial canals
Recommended for use with a guide wire. Cannulated
to allow guide wire use. Features a quick-connect
end for use with a driver.

PRODUCT NO:

2628

Overall Length: 10" (25,4 cm)
Reamer Diameter: 7,5 mm

1.800.548.2362       NOVEMBER 2023   

PRODUCT NO:

8248 [Fixed Driver]with Zimmer Hall Quick-connect   HIP ARTHROPLASTY INSTRUMENTS

5

Lombardi Self-holding X-ray Magnification Marker

The flexible, adjustable arm
can help reduce patient (andtechnologist) embarrassment or
discomfort when it is required to
be positioned in a sensitive areasuch as the inner thigh.

Designed by Adolph Lombardi, MD

Helps to remove the variable of X-Ray magnification
factor from the process of Orthopedic templating
Fully positionable, this orthopedic X-Ray calibration and marking device features a 1"
(25.4mm) stainless steel ball which, when properly positioned at bone level on a precise
anatomical plane, will be this exact size when viewed from all angles, allowing it be used asa calibration marker in surgical planning software applications, helping to gauge the size ofother components on that plane. This helps establish precise anatomical measurement.
PRODUCT NO:

2672

Base Dimensions: 11" x 5.25" (27,9 x 13,3 cm)
Post Height: 7" (17,8 cm)
Arm Maximum Length: 13" (33 cm)

Articulated Measuring Device with Ruler

Designed by Vincent Y. Ng, MD

A highly precise (within 1 mm) device designed for measuring
distances between two points — can be used even if there
are intervening structures like soft tissue or bone, and insituations where a straight ruler will not work
Examples of use include measuring limb length in total hip arthroplasty, confirming
length in megaprosthetic knee replacements, and assessing dimensions of allografts.
PRODUCT NO’S:

2026-00 [Measuring Device with Ruler]
Set Includes/ Available Separately:2026-01 [Measuring Device Only]

Overall Length (unfolded): 15.25" (38,8 cm)
Dimensions Triangle Folded: 4" x 4.25" (10,2 x 10,8 cm)

2026-02 [Ruler Only]

Overall Length: 9" (22,9 cm)
Width: .79" (2 cm)

Kenerly Femoral Neck Cutting Guide
Designed by J. Lex Kenerly, III, MD

Designed for use during the anterior
approach for THA to help determine thefemoral neck osteotomy location, the guideis placed on the femoral neck and adjusted
using the intraoperative C-arm image
to visualize and compare to the pre-op
templating, providing an excellent location
for the initial femoral neck osteotomy
6

PRODUCT NO:

4590

Overall Length: 8.25" (21 cm)
Handle Length: 1.9" (4,8 cm)
Cutting Guide Bar Length: 1.22" (3,1 cm)
End of Bar to Tip Length: 3.5 mm
Shaft Angle at End: 30°
Shaft Diameter .125" (3,2 mm)

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Lombardi Taper Cleaner
Designed by Adolph V. Lombardi Jr., MD

Designed to help clean a hip stem
taper of corrosive by-products prior toplacement of the new femoral head
PRODUCT NO’S:

Overall Length: 2.125" (5,4 cm)
Outside Diameter: 1" (2,54 cm)
8034 Small Short Taper 11.3/12.2 mm
8034-01 Long Taper 11.4/13.4 mm
8035-01 11/13 mm
8035-02 12/14 mm
8035-03 14/16 mm

Kudrna Hip StemTaper Protectors
Designed by James Kudrna, MD

Used to cover and protect the
hip stem taper of a femoral
component — especiallyhelpful in cup revision surgery
PRODUCT NO'S:

1151 [11/13]
1152 [12/14]
1153 [14/16]

11/13 #1151

Tissue Protector

PRODUCT NO’S:

Helps protect tissue when astraight reamer is being used

24 mm

#5480-02

19 mm#5480-01

14/16 #1153

12/14 #1152

Designed to be used when a straight reamer
is being used in a bone canal. Very useful inminimally invasive total hip arthroplasty.

5480-01

Inside Diameter: 19 mm
Overall Length: 6.5" (16,5 cm)
Tube Depth: 3.875" (9,8 cm)

5480-02

Inside Diameter: 24 mm
Overall Length: 6.5" (16,5 cm)
Tube Depth: 3.875" (9,8 cm)

Clear Vision Debris Shield
Designed by R. Barry Sorrells, MD

Provides a degree of restriction fromflying debris or liquid during surgery
Held between the surgical site and the operating
personnel, the shield provides a clear undistorted
view, while helping to protect the patient andpersonnel from possible contamination. The shield is
autoclavable and gas sterilizable in a flat position.
PRODUCT NO:

8031-01

Dimensions: 8" x 10.25" (20,3 cm x 26 cm)(Dimensions do not include the handle)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

7

5 mm

#1426-01

Cheng Screw Removal
and Bone Trephine Set

8 mm#1426-03

Six trephine sizes with reverse thread teeth
designed to help with removal of screws
with minimal bone loss, as well asgathering of core bone samples forbiopsy or core decompression

Designed by Edward Cheng, MD

Cannulated
T-Handle

For Screw Removal

6.5 mm#1426-02

Can be used with the T-handle or with power.

9 mm#1426-05

10 mm#1426-06

11 mm#1426-07

PRODUCT NO’S:

1426-00 [Complete Set with Case]
Set Includes/Available Separately:

1426-01 [5 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-02 [6.5 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-03 [8 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-05 [9 mm Internal Diameter]#1425-14

K-wire not included.

Overall Length: 7.125" (18,1 cm)

1426-06 [10 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

The trephine ends are designed to fit over embedded screws for extraction
with minimal bone loss. Six sizes available — internal diameters of 5 mm, 6.5 mm,
8 mm, 9 mm, 10 mm, and 11 mm. The T-handle allows for precise, controlled use.

1426-07 [11 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1425-14 [Handle Assembly]

Dimensions: 4" x 2" (10,2 cm x 5,1 cm)

For Core Bone Sampling

1025 [Sterilization Case]

Cannulated T-handle and trephines allow use of a standard 1.6 mm (.062") threaded K-wire to help facilitate
grasping and removal of a core bone sample for biopsy or core decompression. Variety of core diameters yields
bone samples of sufficient size for pathology. K-wire not included.

Replacement Part:

1425-14-B-COMP [Handle Retaining Screw]

IHS Inclinometer
Designed by Craig J. Della Valle, MD

PRODUCT NO:

Glass free to allow
sterilization with vacuum!

1326

Dimensions: 4" x 2" (10,2 cm x 5,1 cm)

Magnets along the bottom for hands-free use

Helps to accurately
predetermine angles for
acetabular cup positioning
and insertion—calibrated from0 to 45°, the indicator may
be used on the reamer shaft,the trial cup shaft and the cup
impactor shaft
Designed to allow the surgeon toconsistently and quickly achieve the desired
component position during each step ofacetabular preparation and component
positioning: acetabular reaming, trial
component positioning, and actual
component insertion. Steam sterilizable.

AccuAngle Indicator

Designed by S. David Stulberg, MD, A. Llinas, MD and J. Navas, MD

Helps to accurately predetermine
angles for acetabular cup
positioning and insertion
Calibrated from 0 to 45°, the indicator may be
used on the reamer shaft, the trial cup shaft andthe cup impactor shaft.
Designed to allow the surgeon to consistently andquickly achieve the desired component position
during each step of acetabular preparation andcomponent positioning: acetabular reaming, trial
component positioning, and actual component
insertion. Steam sterilizable without vacuum.

WARNING: Do not strike glass indicator tube.

PRODUCT NO:

1325

Dimensions: 4" x 2" (10,2 cm x 5,1 cm)

8

Magnets along the bottom for hands-free use

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Duellman Total Hip Trunnion Clamp
Designed by Todd Duellman, MD

Designed for use on a trial modular
neck/trunnion at the time of placement
on/off the femoral stem to help
determine offset and neck length

PRODUCT NO:

1817

Overall Length: 8" (20,3 cm)
Clamp End Internal Diameter: 10 mm
Clamp End Width: 11.5 mm

Namba Hip Slide

Helps reduce a femoral head trial and implant into theacetabulum during total hip surgery. Manufactured ofdelrin to help eliminate damage to the implant. Can
be steam or gas sterilized and is radiolucent. Threesizes to accommodate different diameter heads.

Designed by Robert S. Namba, MD

Safely glides femoral heads
into the acetabulum —essential for ceramic heads

Facilitates MIS hipreplacement procedures50-60 mm#6892

40-48 mm#6891

22-40 mm
#6890

Wiater Slide

Smallest size now
accommodates
up to 40 mm

PRODUCT NO’S:

Overall Length: 12" (30,5 cm)

6890 For 22–40 mm heads
6891 For 40–48 mm heads
6892 For 50–60 mm heads

Designed by J. Michael Wiater, MD, FAAOS, FAOA

Useful for total hip arthroplasty or hip preservation
procedures in smaller patients
Also helps avoid damage to the prosthetic bearing
surfaces during dislocation and reductionof a shoulder arthroplasty.

PRODUCT NO:

Manufactured of delrin to help eliminate damage to theimplant. Can be steam or gas sterilized and is radiolucent.

6879

Overall Length: 11" (27,9 cm)
Width: 1.375" (3,5 cm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

9Powers Femoral Sounds

PRODUCT NO’S:

4189-00 [Set of 5]
Also available individually:

Designed by Mark Powers, MD

4189-14 [14 mm]

Overall Length: 14.25" (36,2 cm)
Handle Length: 3.5" (8,9 cm)

4189-12 [12 mm]

Overall Length: 14.25" (36,2 cm)
Handle Length: 3.5" (8,9 cm)

4189-10 [10 mm]

Overall Length: 14.25" (36,2 cm)
Handle Length: 3.5" (8,9 cm)

14 mm#4189-144189-08 [8 mm]

Overall Length: 14.25" (36,2 cm)
Handle Length: 3.5" (8,9 cm)

12 mm#4189-124189-06 [6 mm]

Overall Length: 14.25" (36,2 cm)
Handle Length: 3.5" (8,9 cm)

10 mm#4189-10

8 mm#4189-08

6 mm#4189-06

Allows the surgeon to gently identify
the canal of a long bone as well as its
width (isthmus) prior to inserting a device

Particularly useful for the anterior
approach to the hip. Helps
identify intraoperative occult
fractures. Properly identifying
the medullary canal before
broaching helps minimize
possible intraoperative fractures.

Sanders
Femoral Neck
Cutting Blocks

5 x 10 mm#4575

Designed by Richard A. Sanders, MD

Designed to help with theaccurate placement of thefemoral neck osteotomy intotal hip surgery

10 x 10 mm#4560

10 x 15 mm#4565

Used to measure the distance from theproximal end of the lesser trochanter to thelevel of the femoral neck osteotomy. The
desired level of the femoral neck osteotomy isdetermined by preoperative planning. The exact
level of the femoral osteotomy helps with leg
length, either maintaining equal leg length orcorrecting leg length discrepancies.

10 x 20 mm#4570

10 x 25 mm#4575

PRODUCT NO’S:

4555 [5 x 10 mm]

Overall Length: 6.5" (16,5 cm)
Block: 5 x 10 mm

4560 [10 x 10 mm]

Overall Length: 6.5" (16,5 cm)
Block: 10 x 10 mm

4565 [10 x 15 mm]

Overall Length: 6.5" (16,5 cm)
Block: 10 x 15 mm

4570 [10 x 20 mm]

Overall Length: 6.5" (16,5 cm)
Block: 10 x 20 mm

4575 [10 x 25 mm]

Overall Length: 6.5" (16,5 cm)
Block: 10 x 25 mm

Ruler with 45° Angle Handle
Designed by Richard A. Sanders, MD

Useful for measuring distancesin small deep incisions
Ideal for measuring the distance from thelesser trochanter to the center of the trial
femoral head during femoral sizing.
Prototype
Shown

Ruler with Right Angle Handle
Designed to be used to measure
the femoral head/neck length
Very helpful in minimally invasive surgery.
PRODUCT NO:

PRODUCT NO:

1450

1430

Handle Length: 4.25" (10,8 cm)
Ruler Dimensions: 2.5" x .5" (6,4 cm x 1,3 cm)

Handle Length: 5" (12,7 cm)
Ruler Dimensions: 2.5" x .5" (6,4 cm x 1,3 cm)

10

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Sarraf CoatedHip Dislocation Hook
Designed by Khaled M. Sarraf, MD

Coated hook end helps to prevent from marring component surfaces.
Can also be used as a bone hook, and for femoral elevation.

Designed to aid in dislocatinga femoral stem while helping toprevent damage to the trunion
PRODUCT NO:

5905

Curve Diameter: 50 mm
Overall Length: 12.5" (31,8 cm)
Handle Length: 4.75" (12,1 cm)

Mengato Depth Gauge

PRODUCT NO:

1139

Overall Length – Contracted: 7.125" (18,1 cm)
Overall Length – Extended: 9.125" (23,2 cm)
Gauge: 0 to 50 mm

Designed by Richard Mengato, MD
US Patent# 8,512,349

Ring-handled design with
3 rings gives 3-point grip
for ease of holding andmanipulation
Allows for superior gauge control andmanipulation, to advance, engage andmaintain the hook on the distal cortex bylevering the probe against the bone holeand keeping gentle tension on the hook.

Depth Gauge

Designed for one-handed use —helps to provide measurement of
the depth/length of any bone hole forproper screw length determination

PRODUCT NO:

8015

Overall Length: 7.625" (19,4 cm)
Scale: From 0 to 48 mm

Sterilizable Level

Helpful in hip surgery to ensure the leg is in
the same position when checking leg length
Steam sterilizable without vacuum for use in surgery.
PRODUCT NO:

1180

Dimensions: 2" x .5" x .75" (5,1 cm x 1,3 cm x 1,9 cm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

11

Curved Canal Rasps
Design modification by Michael Messieh, MDof original design by Anthony Unger, MD.

8 mm #3004-01-08

Designed for preparation of
the femoral canal for insertionof a cemented or cementless
hip stem, the mutiple
diameters serve to prepare thefemoral canal after the initial
5 mm is used to find thecurvature of the canal

PRODUCT NO’S:

3004-01-08 [8 mm]

Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)

3004-01-10 [10 mm]

Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)

10 mm #3004-01-10

3004-01-12 [12 mm]

Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)

12 mm #3004-01-12

Unger Canal Finder Rasps

Straight #3004

Designed by Anthony Unger, MD
PRODUCT NO’S:

3004 [Canal Finder Rasp–Straight]
Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)

3004-01 [Canal Finder Rasp–Curved]
Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)

Curved #3004-01

3004-02 [Canal Finder Rasp–Curved     with Smooth Proximal]
Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)

Smooth Proximal Section

Curved with Smooth
Proximal #3004-02

Designed to help shape the femoral canal after reaming

DAA Canal Finder Rasp

Designed to help begin preparationof the femoral canal prior to stem
broaching — features a large
handle with a striking plate end

PRODUCT NO:

C1026

Overall Length: 13.125" (33,4 cm)
Handle Length: 6" (15,2 cm)
Maximum Shaft Diameter: 13 mm

Kim Anterior Total Hip Awl
Designed by William C. Kim, MD

PRODUCT NO:

8028

Overall Length: 12" (30,5 cm)
Blunt Reamer Length: 2" (5,1 cm)

12

Designed to help avoid perforationof the femoral canal while helping
to give an accurate assessment ofcanal orientation for trial broaching
during anterior approach THA

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Offset Femoral Rasp

Designed by Richard Pelliccio

The deep offset design allows the surgeon to line up with canal entry andthe tip angled slightly upwards to help prevent femoral protrusion
PRODUCT NO:

4988

Overall Length: 13.67" (34,7 cm)
Handle Length: 5.88" (15 cm)
Rasp Offset: 1.5" (3,8 cm)

ORIGINAL
DR. ROCKOWITZ DESIGN
Topside Rasp

Rasp on curve topside and sides,
smooth on underside#4990

T-Handle Femoral Canal Finders

Designed to sound the femoral canalprior to stem broaching, especially usefulto help start the broach path during thedirect anterior approach
Rockowitz T-Handle Femoral Canal Finder Rasp

SMOOTH DESIGNPRODUCT NO:

No rasp — smooth underside,
sides, and topside#4990-03

4990

Designed by
Neal L. Rockowitz, MD

Overall Length: 9" (22,9 cm)
Curved Rasp Portion: 4" (10,2 cm)

T-Handle Femoral Canal Finder – Smooth
PRODUCT NO:

4990-03 [Smooth]Overall Length: 9.385" (24,4 cm)

MODIFIED DESIGN
Underside Rasp

Modified T-Handle Femoral Canal Finder Rasp

Rasp on curve underside and sides,
smooth on topside#4989

PRODUCT NO:

4989

Overall Length: 9" (22,9 cm)
Curved Rasp Portion: 4" (10,2 cm)

Kenerly
Double Parallel
Femoral Neck
Disc Grasper
Design modified by J. Lex Kenerly, III, MD

PRODUCT NO:

1720-01

Overall Length: 10" (25,4 cm)
Jaw Dimensions: 1.44" x .72"
 	
(36,6 x 18,3 mm)
Lower Jaw Thickness: 1 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Designed to remove thecentral disc of a double,
parallel cut femoral
neck osteotomy when
performing THA

Durham Curved Osteotome
Designed by Alfred A. Durham, MD

Increased angle useful for posterior
osteophytes of the femoral condyleand the humeral head, as well asanterior acetabular osteophytes
1.800.548.2362       NOVEMBER 2023   

PRODUCT NO:

4950

Overall Length: 9" (22,9 cm)
Handle Length: 5" (12,7 cm)
Osteotome Width: .625" (1,6 cm)   HIP ARTHROPLASTY INSTRUMENTS

13

Wells Modified Lambotte
PAO Osteotomes

PRODUCT NO’S:

5276-00 [Set with case]

Straight PAO Osteotome #5276-01

Set Includes / Available Individually:

5276-01 [Straight]

Overall Length: 14" (35,6 cm)
Handle Length: 4.5" (11,4 cm)
Ostetome Width: 11,1 mm

Designed by Joel Wells, MD

Designed to focus on theposterior column osteotomy
and connection to the ischial
cut — straight, curved andtwo offset options helps theposterior column osteotomy
to be cut with more control

Curved PAO Osteotome #5276-02

5276-02 [Curved]

Overall Length: 13.875" (35,3 cm)
Handle Length: 4.5" (11,4 cm)
Ostetome Width: 12,7 mm

5276-03 [Offset]

Offset PAO Osteotome #5276-03

Overall Length: 13.625" (34,6 cm)
Handle Length: 4.5" (11,4 cm)
Ostetome Width: 12,7 mm

5276-04 [Offset Curved]
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Silicone handle designed for better control.

Offset Curved PAO Osteotome #5276-04

Overall Length: 13.625" (34,6 cm)
Handle Length: 4.5" (11,4 cm)
Ostetome Width: 12,7 mm

9007 [Case]

Modified Lambotte Osteotomes

Designed with a striking platform, plus across-bar hole to help control rotational
stability and assist with removal
PRODUCT NO’S:

5350-00 [Set w/Case]* MADE EXCLUSIVELY

Also Available Individually:

GERMANY

5350-25* [1/4"]

Six (6) sizes available, from 1/4" to 1-1/2"
in 1/4" increments. Cross-bar and caseincluded in complete set. Two smallest sizes
have an 1/8" hole in which an 1/8" pin canbe used as a cross bar (not included).

FOR INNOMED IN

Overall Length: 9" (22,9 cm)
5350-125 [1-1/4"]
Osteotome Width: .25" (6,4 mm)
Overall Length: 9" (22,9 cm)
Osteotome Width: 1.25" (31,8 mm)
5350-50* [1/2"]
Overall Length: 9" (22,9 cm)
5350-150 [1-1/2"]
Osteotome Width: .5" (12,7 mm)
Overall Length: 9" (22,9 cm)
Osteotome Width: 1.5" (38,1 mm)
5350-75 [3/4"]
Overall Length: 9" (22,9 cm)
5350-CASE [Case]
Osteotome Width: .75" (19 mm)
Dimensions: 12.25" x 11.25" x 1"
(31,1 x 28,6 x 2,5 cm)
5350-100 [1"]Overall Length: 9" (22,9 cm)
5350-CB [Cross Bar]
Osteotome Width: 1" (25,4 mm)
Overall Length: 4.375" (11,1 cm)

Lambotte Osteotomes with Handle
Designed by John Cherf, MD

Handle allows for better control,
reducing rotation
during use
PRODUCT NO’S:

5250-01 [Straight]

Blade Width: .25" (6.3 mm)
Overall Length: 13" 32,8 cm)
Handle Length: 4.5" (11,4 cm)

5260-01 [Curved]

Blade Width: .25" (6.3 mm)
Overall Length: 13" 32,8 cm)
Handle Length: 4.5" (11,4 cm)

Wagner Osteotome Handle
Handle designed by Russell Wagner, MD

Handle is designed for easiergripping, rotational control, anduse with a mallet with a standard
1/4” Lambotte osteotome

Straight #5250-01
Curved #5260-01

Designed with a handle for better control, which helps reduce rotation of theosteotome during use. The handle also provides a larger striking area for use with a mallet.
The osteotome shafts are manufactured with stainless steel and are available both straight and curved.

Osteotome #5348-01

Handle #5348
PRODUCT NO'S:

5348 [Handle Only]

Overall Length: 5.5" (14 cm)

5348-01 [1/4" Osteotome Only]
Overall Length: 8.875" (22,5 cm)

14

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Mueller Style Hip Instruments
PRODUCT NO’S:

6865-01 [Flat Blade Osteotome]
Overall Length: 11.125" (28,3 cm)
Osteotome Width: 20 mm

20mm
Flat Blade Osteotome

6865-02 [Femoral Head Dislocation Lever]

16mm
Wide Curved Osteotome#6865-01

Overall Length: 11.375” (23,8 cm)
Scoop Dimensions: 25 mm x 57 mm#6865-04

6865-03 [Narrow Curved Osteotome]
Overall Length: 12” (30,5 cm)
Osteotome Width: 9 mm

6865-04 [Wide Curved Osteotome]

25mm
Femoral Head Dislocation Lever

Overall Length: 12” (30,5 cm)
Osteotome Width: 16 mm

23mm
Swan Neck Curved Gouge#6865-02

6865-05 [Swan Neck Curved Gouge]#6865-05

Overall Length: 12” (30,5 cm)
Gouge Width: 23 mm

5350-CB [Cross Bar]

Cross Bar #5350-CB

9mm
Narrow Curved Osteotome#6865-03

PRODUCT NO’S:

5270-04

Mini-lexer Osteotomes

Blade Width: 12 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

Helpful in osteophyte andcement removal
Small, thin osteotomes helpful inosteophyte and cement removal in
total joint surgery. Larger handle
helps with better control.
MADE FOR INNOMED IN

GERMANY

5270-03

Blade Width: 10 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

12 mm #5270-04

5270-02

Blade Width: 6 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

10 mm #5270-03

5270-01

6 mm #5270-03

Blade Width: 4 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

4 mm #5270-01

Sarraf Toothed Curettes

Forward, straight, and reverse bent toothed curettes
designed to aid in aid in all types of joint arthroplasty
surgery, especially in scraping any articularchondral islands within the acetabulum
during THA preparation

Designed by Khaled Sarraf, MD
PRODUCT NO’S:

5174-00 [Set]
Set Includes/ Available Separately:

5174-F [Forward Toothed Curette]
Overall Length: 11.5" (29,2 cm)
Handle Length: 5.5" (14 cm)
Curette Cup : 8 mm X 12 mm
Angled Down: 30°

5174-R [Reverse Toothed Curette]
Overall Length: 11.5" (29,2 cm)
Handle Length: 5.5" (14 cm)
Curette Cup : 8 mm X 12 mm
Angled Up: 30°

5174-S [Straight Toothed Curette]
Overall Length: 11.5" (29,2 cm)
Handle Length: 5.5" (14 cm)
Curette Cup : 8 mm X 12 mm

Forward Toothed
Can also be used for the femoral
canal in cemented and uncemented THA
Valuable aid in revision arthroplasty (hip, knee,shoulder and ankle) for cement curettage
Useful tool in hip and knee primary arthroplasty as well
as shoulder, elbow and ankle arthroplasty procedures
Ultra hard titanium nitride coating helps to extend life by increasing surface
hardness, prolonging sharpness, and resisting chemicals and corrosion.

#5174-F

Reverse Toothed#5174-R

Straight Toothed#5174-S

Chandran Bent Serrated Curette
Designed by Rama E. Chandran, MD

PRODUCT NO:

5171

Overall Length: 11.75" (29,8 cm)
Handle Length: 5.5" (14 cm)
Cup Size: 7 mm X 12 mm

Serrated design allows for easier removal of cancellous
bone in the proximal femur in total joint arthroplasty
Ultra hard titanium nitride coating helps to extend life by increasing surface
hardness, prolonging sharpness, and resisting chemicals and corrosion.

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

15

Large Bone Curettes

Angled Small #5160-01

Designed with a 5/16" (8 mm)
diameter shaft allowing better
visualization into the medullary
canalStraight Small #5160-02
Angled Medium #5160-03

PRODUCT NO’S:

5160 [Set with Case]

Angled Large #5160-04

Individual Instrument Dimensions:Overall Length: 15" (38,1 cm)
Handle Length: 4.5" (11,4 cm)

5160-01 [Angled Small]
Curette End: 10 mm X 18 mm

Straight Medium #5160-05

5160-02 [Straight Small]
Curette End: 10 mm X 18 mm

5160-03 [Angled Medium]
Curette End: 10 mm X 24 mm

5160-04 [Angled Large]
Curette End: 24 mm X 24 mm

5160-05 [Straight Medium]

Curette End: 10 mm X 24 mm

9007 [Case Only]

Large

Medium

Small

24 mm
x
24 mm

10 mm
x
24 mm

10 mm
x
18 mm

Curette Ends
at Actual Size

PRODUCT NO’S:

5190-00 [Set of Three]Powers Double Bent Curette Set

Also available individually:

5190-L [Angled Left]

Overall Length: 16.875" (42,9 cm)
Handle Length: 9" (22,9 cm)
Shaft Length Before Bend: 5.25" (13,3 cm)
Bend Offset: .5" (1,3 cm)
Curette Cup Angle: 33°
Curette Cup Inner Dimen.: 6 mm X 8,7 mm

Designed by Mark Powers, MD

The bayonet curettes help allow for proper lateralization and seating of the broach
Angled Left #5190-L

5190-S [Straight]

Overall Length: 17" (43,2 cm)
Handle Length: 9" (22,9 cm)
Shaft Length Before Bend: 5.25" (13,3 cm)
Bend Offset: .5" (1,3 cm)
Curette Cup Angle: 33°
Curette Cup Inner Dimen.: 6 mm X 8,7 mm

Straight #5190-S

5190-R [Angled Right]

Overall Length: 16.875" (42,9 cm)
Handle Length: 9" (22,9 cm)
Shaft Length Before Bend: 5.25" (13,3 cm)
Bend Offset: .5" (1,3 cm)
Curette Cup Angle: 33°
Curette Cup Inner Dimen.: 6 mm X 8,7 mm

Angled Right #5190-R

Ring Curettes – Straight Shaft
3 mm Straight #5150

3 mm Bent #5156

6 mm Straight #5152

6 mm Bent #5157

8 mm Straight #5154

8 mm Bent #5158

PRODUCT NO’S:
Straight Shaft

Overall Length: 8.75" (22,2 cm)

5150 [3 mm Straight]
Ring Diameter: 3 mm

16

Ring Curettes – Bent Shaft

PRODUCT NO’S:
Bent Shaft

MADE FOR INNOMED IN

GERMANY

5152 [6 mm Straight]

Ring Diameter: 6 mm

5154 [8 mm Straight]
Ring Diameter: 8 mm

MADE FOR INNOMED IN

Overall Length: 8.625" (21,9 cm)

5156 [3 mm Bent]

Ring Diameter: 3 mm

INNOMED.NET   GERMANY

5157 [6 mm Bent]

Ring Diameter: 6 mm

5158 [8 mm Bent]

Ring Diameter: 8 mm   FREE TRIAL ON MOST INSTRUMENTS

Rivero Anti-Rotation Corkscrew Femoral Head Remover

PRODUCT NO’S:

3705 [Corkscrew & Sleeve Set]
Overall Length: 10" (25,4 cm)

Designed by Dennis Rivero, MD

Individual Instruments:

Designed to help prevent rotation while
engaging a femoral head for removal

3705-01 [Corkscrew Only]
Overall Length: 10” (25,4c m)

3705-02 [Sleeve Only]

Overall Length: 8” (20,3 cm)

The sharp-toothed sleeve can be tapped in to help provide
purchase of the femoral head, then held to help prevent
rotation as the super-threaded corkscrew is turned toengage the head for removal.

STABILIZE HEAD

INSERT/ENGAGE CORKSCREWRivero Extra Grip
Femoral Head Removers

REMOVE HEADPRODUCT NO:

8248 [Fixed Driver]with Zimmer Hall Quick-connect

Modified by Dennis Rivero, MD

Used to remove femoral heads during
total hip arthroplasty or fracture surgery

Zimmer Hall Quick-connect#3706

Quick-connect version for use with a driver.
PRODUCT NO’S:

3706 [Zimmer Hall Quick-connect]
Overall Length: 8.5” (21,6 cm)

T-Handle#3707

3707 [T-Handle]

Overall Length: 8.75” (22,2 cm)

Femoral Head Removers

Extra Gripping Thread

PRODUCT NO:

8248 [Fixed Driver]with Zimmer Hall Quick-connect

Used to remove a femoral head during
total hip arthroplasty or fracture surgery
Quick-connect version for use with a driver.
PRODUCT NO’S:

Zimmer Hall Quick-connect

3688 [Zimmer Hall Quick-connect]#3688

Overall Length: 8.5" (21,6 cm)

3690 [T-Handle]

Overall Length: 8.75" (22,2 cm)

T-Handle#3690

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

17

Huddleston
Femoral Head Removers
Designed by H. Dennis Huddleston, MD

Designed to help lever a femoral
head out of the acetabulum instandard and anterior approach
total hip replacement
PRODUCT NO'S:

3608 [Sharp]

Overall Length: 10.5" (26,7 cm)
Scoop Length: 3" (7,6 cm)
Scoop Width: 29 mm

3609 [Dull]

Overall Length: 10.5" (26,7 cm)
Scoop Length: 3" (7,6 cm)
Scoop Width: 29 mm

Sharp #3608
Dull #3609

Femoral Head Removal Clamp
Firmly locks onto a resected
femoral head during total hip, hipfracture, and MIS total hip surgery

Designed to firmly lock onto a resected femoral
head during total hip surgery or hip fracture. Narrowdesign is also useful in minimally invasive total hip
surgery with limited access to the femoral head.
PRODUCT NO:

3680

Overall Length: 10.75" (27,3 cm)

Verner Corkscrew
Femoral Head Remover

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Used to remove the femoral head
during total hip arthroplasty or fracture surgery

Designed by James J. Verner, MD & Andy Lytle

Designed so the threads engage the head
under power and draws the corkscrew in until
the head begins to turn.
The extra long shaft keeps the power
reamer out of the operative site for better
visualization and improves the lever arm when
pivoting the head out of the acetabulum. The
grip ring allows the surgeon to pull head outof acetabulum and soft tissue envelope whendisengaged from the driver.
Features a Zimmer Hall Quick-connect for
use with a driver.

PRODUCT NO:

8248 [Fixed Driver]with Zimmer Hall Quick-connect

PRODUCT NO:

3698

Overall Length: 12.25" (31,1 cm)

18

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

O’Reilly Femoral Head Extractor
Designed by Michael P. O’Reilly, MD
Small version designed modification by Tarum Bhargava, MD

Designed to help remove the femoral head
during THA, MIS Direct Anterior THA, and hip
fracture surgery/hemiarthroplasty
The perpendicular osteotome blades help provide purchasein osteoporotic bone, while the central osteotome provides avisual estimate of the instrument’s depth of penetration to avoid
acetabular injury with use during hemiarthroplasty.
The handle helps obtain rotational torque needed to rotateand dislocate the femoral head in direct anterior hip arthroplasty.

Large #3675

PRODUCT NO’S:

3675 [Large]

Overall Length: 9.5" (24,1 cm)
Hammer Platform Diameter: 1.125" (2,9 cm)
Width at End: 1.1" (2,8 cm)

Small #3674

3674 [Small]

Overall Length: 9.5" (24,1 cm)
Hammer Platform Diameter: 1.125" (2,9 cm)
Width at End: .75" (1,9 cm)

Schantz Pin with Zimmer Hall Quick-connect
Designed by Keith Berend, MD

Used to help remove a femoral head during total hip surgery

PRODUCT NO:

3687

Overall Length: 8.625" (21,9 cm)
Shaft Length: 7.375" (18,7 cm)
Thread Length: 2.5" (6,4 cm)
Diameter: 4.5 mm

Partial threaded pin can be used to help remove a femoral head during total hip
surgery. Especially helpful in minimally invasive total hip surgery where access to thefemoral head is limited. Connects with a Zimmer Hall Quick-connect.

Femoral Head Removal Pin

Used to help remove a femoral head during total hip surgery
Partial threaded pin can be used to help remove a femoral head during total hip surgery. The pin is especially helpful inminimally invasive total hip surgery where access to the femoral head is limited. The pin is attached to a pin driver whichclamps onto a Jacob chuck. When the pin is drilled in place, the driver is easily removed from the pin, as the pin is heldby a friction ring. The head can be removed by gripping the pin by hand or by using a large pin inserter/extractor.

Designed to be used with:

Pin Driver #1205

PRODUCT NO’S:

1310 [Pin]Overall Length: 9" (22,9 cm)
Diameter: 5/32" (4 mm)

Optional Inserter/Extractor:

1205 [Pin Driver]1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

19

Universal Bone
Grafting/Impacting
ForcepsBone graft can be grasped, placed& impacted without changing
hands or instruments

Designed by J. A. Amis, MD

The forceps are designed withgrasping ends for delivery of bone
graft. When the graft is in place,the forceps are closed, whichforms the ends into an impacting
punch. A striking platform is
attached to the end of theforceps for tapping and tamping
the graft. Four end diameters areavailable in two lengths.
PRODUCT NO’S:
Short: 6" (15,2 cm) Length

5010-01 1/8" (3,2 mm) Diameter End
5010-02 3/16" (4,8 mm) Diameter End
5010-03 1/4" (6,3 mm) Diameter End
5010-04 5/16" (8 mm) Diameter End

Long: 10" (25,4 cm) Length

5050-01 1/8" (3,2 mm) Diameter End
5050-02 3/16" (4,8 mm) Diameter End
5050-03 1/4" (6,3 mm) Diameter End
5050-04 5/16" (8 mm) Diameter End

1/8"
3/16"
1/4"
5/16"
(3,2 mm) (4,8 mm) (6,3 mm)(8 mm)
Diameter ends at actual size (closed forceps)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

When the forceps
are closed, they
form into an
impacting punch

Double Ended Grater Cleaning Tool
Designed by Brandon Thompson, CST/CFA

Acetabular grater bone remover
PRODUCT NO:

8007

Overall Length: 7" (17,8 cm)

Allograft Bone Vise

Holds allograft bone forreaming, shaping or cutting
The vise is designed with two sets of visejaws for reaming of two femoral heads andalso for holding a long bone horizontally andvertically. The base plate is designed with atable flange for stabilization during use. The
vise is completely autoclavable.
PRODUCT NO:

8215

Base Dimensions: 8.25" x 11" (21 cm x 27,9 cm)

20

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

9 mm Round

12 mm Round#5337

15 mm Round

#5336

#5335

9 mm Square

12 mm Tapered#5334

12 x 7 mm Rectangle

#5333

#5332

11 x 4 mm Rectangle#5331

Ortho Impactors
PRODUCT NO’S:

Overall Length: 9" (22,9 cm)
Shaft Diameter: 9 mm

5337 [9 mm Round]5336 [12 mm Round]5335 [15 mm Round]5334 [9 mm Square]5333 [12 mm Tapered]5332 [12 x 7 mm Rectangle]
5331 [11 x 4 mm Rectangle]

Stainless

Impactor Sizes

DelrinImpactor Sizes

Modular Impactor Set

Makes multiple impactor heads
easily visible and available
Designed to have available to the operating
surgeon multiple types of impactors utilizing one
handle. The rack uses less space and allows thesurgeon to quickly see the designs available. The
impactors are supplied with stainless steel tips forbone and delrin tips which can be used against
an implant for slight placement adjustments.

PRODUCT NO:

5370 [Complete Set]
Included In Set / Also Available Individually:

5370-01 [Rectangular Tip 11 mm x 4 mm Steel]
5370-02 [Oval Tip 13 mm x 8 mm Steel]
5370-03 [Crescent Tip 12 mm x 5 mm Steel]
5370-04 [Square Tip 9 mm x 9 mm Steel]
5370-05 [Round Tip 15 mm Steel]
5370-06 [Round Tip 12 mm Steel]
5370-07 [Round Tip 9 mm Steel]
5370-19 [Set Base] Base Diameter: 3.5" (8,9 cm)
5370-D1 [Rectangular Tip 11 mm 4 mm Delrin]
5370-D2 [Oval Tip 13 mm x 8 mm Delrin]
5370-D3 [Crescent Tip 12 mm x 5 mm Delrin]
5370-H [Modular Handle]
Overall Length: 8" (20,3 cm)
Grip Length: 4.5" (11,4 cm)

Bone Graft Impactors

Tap bone graft or bone parts into
place with minimal bone trauma
PRODUCT NO’S:

5310 [Round]

Head Diameter: 12.5 mm
Overall Length: 9.5" (24,1 cm)
Handle Length: 4.25" (10,5 cm)

5320 [Square]

Head Dimensions: 10 mm x 10 mm
Overall Length: 9.5" (24,1 cm)
Handle Length: 4.25" (10,5 cm)

5325 [Square with Delrin Tip]
Head Dimensions: 10 mm x 10 mm
Overall Length: 9.5" (24,1 cm)
Handle Length: 4.25" (10,5 cm)

Designed with serrated,stainless steel tips andavailable in three shapes:
round, square and rectangular.

Rectangular

5330 [Rectangular]

Head Dimensions: 10 mm x 3 mm
Overall Length: 9.5" (24,1 cm)
Handle Length: 4.25" (10,5 cm)

1.800.548.2362       NOVEMBER 2023   

Square

Round

#5320

Square with
Delrin Tip#5310   HIP ARTHROPLASTY INSTRUMENTS

#5330

#5325

21

Namba Bone Graft Slide
Designed by Robert S. Namba, MD

Helps to efficiently guide allograft
material into the acetabulum
Helps reduce waste of expensive allograft
material by providing a holding trough andslide for effective, directed delivery.
PRODUCT NO’S:

6888

Overall Length: 7.75" (19,7 cm)

Desai Surgical Funnel

Designed by Sarang Desai, DO

Helps with control and placement of bone graft
Made from surgical grade stainless steel (for sterilization).
Profile View

PRODUCT NO:

8989

Overall Length: 6.25" (15,9 cm)
Handle Length: 3.25" (8,3 cm)
Funnel Diameter at Top: 3" (7,6 cm)
Funnel Flow-thru Diameter: 11 mm

Lombardi Cement/Antibiotic Sifter
Designed by Adolph V. Lombardi Jr., MD

PRODUCT NO:

5215

Overall Length: 14" (35,6 cm)
Sifter Diameter: 5" (12,7 cm)

Surgical Spoon
Designed by David Scott, MD

Very useful for the application of
methylmethacrylate bone graft
Made from surgical grade stainless steel(for sterilization purposes).
PRODUCT NO:

8209

Overall Length: 5.875" (14,9 cm)

22

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Soft Tissue
Protector

Cheng Biopsy Trephine System
Three trephine tip sizes

Trephine
DilatorK-wireWire
Repositioning
Guide Tip

Designed by Edward Cheng, MD

Cannulated T-handle and trephines
allow use of a standard 1.6 mm (.062")
threaded K-wire to help facilitate grasping
and removal of a core bone sample forbiopsy or core decompression
Allows use of trephine at oblique angles to bone surface byusing an anchoring K-wire and cannulated trephine
Avoids “skipping” of trephine teeth on bone surface
Facilitates optimal approach angle and direction of trephine
Variety of core diameters yields bone samples of sufficient sizefor pathology
Adapters allow for use of a power drill
Minimally invasive — soft tissue sleeve protects surrounding
structures and tissue
Can also be used for bone graft harvesting
Repositioning guide allows easy adjustment of targeting K-wireTrephines(3 Sizes)

Dilators(4 Sizes)

PRODUCT NO’S:

1425-00 [Complete Set with Case]
Set Includes/Available Separately:

1425-01 [Soft Tissue Protector – Small]
1425-02 [Soft Tissue Protector – Medium]
1425-03 [Soft Tissue Protector – Large]
1425-04 [Dilator – 4.75 mm]
1425-05 [Dilator – 6.25 mm]
1425-06 [Dilator – 7.75 mm]
1425-07 [Dilator – 9.25 mm]
1425-08 [Trephine – Small]

Wire
Repositioning
Guide

Internal Diameter: 5mm
Overall Length: 7.125" (18,1 cm)

1425-09 [Trephine – Medium]

DriverRetraction
HandleTrephine
Drivers

Internal Diameter: 6.5 mm
Overall Length: 7.125" (18,1 cm)

SoftTissueProtectors

1425-10 [Trephine – Lage]
Internal Diameter: 8 mm
Overall Length: 7.125" (18,1 cm)

1425-11 [Drive End – Small]
1425-12 [Drive End – Medium]
1425-13 [Drive End – Large]
1425-14 [Driver Retraction Handle](3 Sizes)

Complete Set in Case

(3 Sizes)

Includes (2) Handle Retaining Screws (#1425-14-B-COMP)

1425-15 [3-Hole Wire Repositioning Guide]1425-Case [Case]
Replacement Part:

1425-14-B-COMP [Handle Retaining Screw]

K-wire not included.

Doroodchi Coated Femoral
Neck Mating Guide
Designed by Hamidreza Doroodchi, MD

Designed for controlled manipulationof femoral head/neck mating inSuperPATH THA approach

PRODUCT NO:

3419

Overall Length: 11.75 (29,8 cm)
Blade Width: 1.125" (29 mm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

23

Bhargava Modular Offset Cup Liner Impactor
Designed by Tarun Bhargava, MD

Designed to help impact an acetabular cup liner during
minimally invasive direct anterior and MIS posterior approach THR
Used in conjunction with individual interchangeable heads(sold separately) which fit securely onto the impactor end
Helps avoid edge loading and improper seating of the linerthat can occur with a straight impactor
Uses the same heads as the Innomed CupX
Actetabular Cup Extraction System

Interchangeable
Head(s) Sold
Separately

PRODUCT NO:

5031

Overall Length: 15.85" (40,2 cm)
Platform End Diameter: 1" (2,54 cm)

Curved Femoral Head Impactor
Designed by Amiee Zirpel

Allows for in-line femoral head impaction
during minimally invasive THR
The curved offset handle allows the head impactor to be
slid under the skin of a small incision, and helps provide
hand-held stability and maneuverability within the wound,while the impaction platform is easily accessible outside
the wound. The impaction disc is made of delrin, whichhelps prevent marring and scratching of components.

PRODUCT NO:

3644

Overall Length: 7.25" (18,4 cm)

Modular Head Holder

Designed by Byron E. Dunaway, MD & Wayne Goldstein, MD

7" #8290-01
9" #8290-02

Designed to hold 22 mm to 36 mm heads forease of insertion in minimally invasive THR

PRODUCT NO’S:

8290-01 [7"]Overall Length: 7" (17,8 cm)

Head holding ends are plastic coated to help eliminate any damage
to the implant. Available in two lengths. Steam and gas sterilizable.

8290-02 [9"]Overall Length: 9" (22,9 cm)

Taper Head Impactor

Designed by Byron E. Dunaway, MD & Wayne Goldstein, MD

Designed to impact a modular
head during minimallyinvasive THR
The impactor has a protective coating to interface against the
implant to help prevent damage while seating the implant. Can
be used with 22 mm to 36 mm heads. Steam and gas sterilizable.

24

INNOMED.NET   PRODUCT NO:

7840

Overall Length: 12" (30,5 cm)   FREE TRIAL ON MOST INSTRUMENTS

Bhargava DAA Femoral Stem Impactor
Designed by Tarun Bhargava, MD

Helps allow for easier impaction of most femoral stems
through the DAA approach — protects the trunion andhelps allow for control of version during impaction

PRODUCT NO:

5308

Overall Length: 10" (25,4 cm)

PRODUCT NO:

5475-10

Diameter: 8 mm
Overall Length: 12.75" (32,4 cm)   Handle Length: 4.75" (12,1 cm)
Shaft Length: 8" (20,3 cm)

Extended Cup Positioner

Designed modification by James F. Kayvanfar, MD of an original design by Thomas Eickmann, MD

Designed to help reposition an acetabular cup during total hip arthroplasty
Ultra hard titanium nitride coating helps to extend life by increasing surface
hardness, prolonging sharpness, and resisting chemicals and corrosion.

PRODUCT NO:

4159

Overall Length: 11.5" (29,2 cm)
Shaft Offset: 1" (2,54 cm)
Head Diameter–Inside: 1.18" (3 cm)
Head Diameter–Outside: 1.5" (3,8 cm)

Blair Acetabular Cup Positioner
Designed by Christopher Blair, DO

Designed to help adjust theposition of an acetabular cup

Offset Cup Liner Inserters

Offset to improve visualization and for mis hip surgery
PRODUCT NO’S:

5032 [32 mm]

32 mm #5032
36 mm #5036

Head Diameter: 32 mm
Overall Length: 16.25" (41,3 cm)

5036 [36 mm]

Head Diameter: 36 mm
Overall Length: 16.25" (41,3 cm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

25

Modified Rongeur with
Pistol Grip Handle

Design modification by Morteza Meftah, MD and Ira Kirschenbaum, MD,of an original design by James T. Mazzara, MD.

A thin top cutter and deep lower cutter, withedges that are rounded off, allows the top
cutter to slide into a tight space— specificallythe acetabulum or the patella—while the pistol
grip helps lessen hand fatigue and slippage,
and allows for better visualization
PRODUCT NO:

1765

Jaw Bite Length: 18 mm
Jaw Bite Width: Tapered from 7 to 4.5 mm
Overall Length: 10" (25,4 cm)

Extra Long Rongeur

Helpful in minimally invasive total hip surgery
by keeping hands out of the field of view

MADE FOR INNOMED IN

GERMANYPRODUCT NO’S:

1771-01

Jaw Bite: 5 x 16 mm
Overall Length: 14" (35,6 cm)

1771-02

Jaw Bite: 8 x 16 mm
Overall Length: 14" (35,6 cm)

1771-03

Jaw Bite: 12 x 16 mm
Overall Length: 14" (35,6 cm)

5 x 16 mm#1771-01

8 x 16 mm#1771-02

12 x 16 mm#1771-03

Mazzara Pistol Grip
Extra Long Rongeur
Designed by James T. Mazzara, MD

Pistol Grip handle lessens
hand fatigue and slippage, andallows for better visualization
PRODUCT NO:

1768-02

Jaw Bite: 7 x 16 mm
Overall Length: 12.5" (31,8 cm)
Shaft-to-End Length: 6" (15,2 cm)

26

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Hannum Modified Angled Grasper
Designed by Scott Hannum, MD

Heavy duty large bone grasper designed
to help trim acetabular osteophytes —angled to ergonomically fit around therim via the direct anterior approach
PRODUCT NO:

1775-04

Overall Length: 8.5" (21,6 cm)
Jaw Width: 11 mm
Jaw Bite Internal: 9 mm x 21 mm

Ortho Rongeur with Easy Grip Handle

Offset handle lessens hand fatigue and slippage,
and allows for better visualization
Offset handle gives better gripping power and helps reduce hand fatigue.
Finger grooves help to prevent hand slippage. The offset handle also
allows for better visualization. Available in three jaw bite sizes.

PRODUCT NO’S:

1780-01

Jaw Bite: 5 x 14 mm
Overall Length: 8.75" (22,2 cm)

1780-02

Jaw Bite: 7 x 16 mm
Overall Length: 8.75" (22,2 cm)

1780-03

Jaw Bite: 10 x 16 mm
Overall Length: 8.75" (22,2 cm)

5 x 14 mm#1780-01

7 x 16 mm#1780-02

10 x 16 mm#1780-03

Mazzara Rongeur
with Pistol Grip Handle
Designed by James T. Mazzara, MD

PRODUCT NO’S:

Pistol Grip handle lessens handfatigue and slippage, and allows
for better visualization

1765-01

Jaw Bite: 5 x 14 mm
Overall Length: 10" (25,4 cm)

1765-02

Jaw Bite: 7 x 16 mm
Overall Length: 10" (25,4 cm)

1765-03

Jaw Bite: 10 x 16 mm
Overall Length: 10" (25,4 cm)

1.800.548.2362       NOVEMBER 2023   

5 x 14 mm#1765-01

7 x 16 mm#1765-02   HIP ARTHROPLASTY INSTRUMENTS

10 x 16 mm#1765-03

27

Tissue Graspers
with Shark Teeth

Up Angled Jaw #1784-01

Designed by Luis Ulloa

Shark teeth help to grasp
on to tissue and bone

Straight Jaw #1784-02

PRODUCT NO’S:

1784-01 [Up Angled Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

1784-02 [Straight Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

Down Angled Jaw #1784-03

1784-03 [Down Angled Jaw]

Shaft allows for use in narrow spaces
Ideal for removing herniated disc material

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

Extra Long Grasper

Designed for reaching deep
into the medullary canal
PRODUCT NO:

1782

Overall Length: 15" (38,1 cm)

Bhargava Anterior Hip Labral Grasper

PRODUCT NO:

1776

Overall Length: 12.5" (31,8 cm)
Shaft Length: 9" (22,9 cm)
Shaft Width: 7 mm
Jaw Width at End: 4 mm
Toothed Jaw Length: 14 mm

Designed by Tarun Bhargava, MD

Designed to help remove the labrum andsoft tissues in anterior total hip surgery, andvery useful in helping to remove posterior
osteophytes in knee surgery

Soudry Loose Body Grasper
Designed by Michael Soudry, MD

Designed to help with the removal of soft tissue
loose bodies in arthroscopy and open procedures
PRODUCT NO:

1769

Overall Length: 9" (22,9 cm)
Shaft Length: 6" (15,2 cm)

28

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYINNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Shark Tooth Graspers
Designed by Luis Ulloa

Helpful in removing the labrum, and osteophytes around the acetabulum andaround the glenoid. Also helps to remove meniscus, osteophytes and loose bodies.
Helps facilitate working through a small incision without disrupting vision.

Sharp teeth help grasp
onto tissue and bone

9" Small Grip #1799-SG

Small Grip Handle
7" Small Grip #1798-SG

PRODUCT NO’S:

1798-SG [7" Small Grip]

Designed with the grip closer
together for easier gripping andto help reduce hand fatigue

Jaw Size: 6 mm x 10 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

1799-SG [9" Small Grip]
Jaw Size: 6 mm x 10 mm
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

12" Standard Grip #1796

Standard Handle
9" Standard Grip #1799

PRODUCT NO’S:

1797 [5" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 8" (20,3 cm)
Shaft Length: 5" (12,7 cm)

7" Standard Grip #1798

1798* [7" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

1799* [9" Standard]5" Standard Grip #1797

Jaw Size: 6 mm x 10 mm
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

1796 [12" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 15" (38,1 cm)
Shaft Length: 12" (30,5 cm)* MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Cartilage Graspers

8" Shark Teeth #1779

Designed by Luis Ulloa

Helps to grasp
and hold cartilage,
tendons, soft tissues
and loose bodies

5" Shark Teeth #1777

Shark Tooth Jaw

Shaft allows for use in
narrow spaces.

6" Saw Teeth #1785
PRODUCT NO’S:PRODUCT NO:

1785 [Saw Teeth]

Shaft Length: 6" (15,2 cm)
Overall Length: 9.25" (23,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       NOVEMBER 2023   

1777 [5" with Shark Teeth]
Shark tooth design modification by Michael Soudry, MD
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Shaft Length: 5" (12,7 cm)
Overall Length: 8.25" (21 cm)
Jaw Bite: 2 mm x 6.5 mm

Saw Tooth Jaw

1779 [8" with Shark Teeth]
Shaft Length: 8" (20,3 cm)
Overall Length: 11.25" (28,6 cm)   HIP ARTHROPLASTY INSTRUMENTS

29

Hannum Tissue Grasper

PRODUCT NO’S:

1775-01 [Short Jaw]

Designed by Scott Hannum, MD

Teeth in jaw firmly holds bone and tissue

Jaw widths (at actual size)
5 mm

8 mm

3 mm

Non-locking design can be easily gripped while allowing
greater pressure to be applied.
Used for dissection (to preserve)/or removal of theanterior capsule, removal of the labrum, or other soft
tissue around the acetabulum prior to cup implantation.
Also used to release the capsule to expose the femur
for placement of the femoral stem. Long, low
profile helps facilitate working through a small
incision without disrupting vision.
Three jaw sizes: short for holding bone, medium
for smaller bones, and long for tissue.

8 mm Jaw Width
Overall Length: 9.25" (23,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1775-02 [Medium Jaw]

5 mm Jaw Width
Overall Length: 9.25" (23,5 cm)

1775-03 [Long Jaw]

3 mm Jaw Width
Overall Length: 9.25" (23,5 cm)

PRODUCT NO’S:

Tapered Jaw #1813

Powers Modified Kocher Clamps

Designed by Mark Powers, MD

1813 [Tapered Jaw]

Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 5.2 mm x 4.1 mm

1813-01 [Tapered Narrow Jaw]
Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 5.2 mm x 3 mm

1814 [Square Jaw]

Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 6.5 mm x 5 mm

Tapered Narrow Jaw #1813-01

Square Jaw #1814

Heavier design allows for a firmer
grasping of bone and soft tissues

Delrin Insert Pliers
Designed to grasp an
implant for adjustment
without marring theimplant surface
PRODUCT NO’S:

2025

Overall Length: 8" (20,3 cm)

2025-03 [Replacement Insert]
Includes top and bottom delrin jaws,
two screws and a hex wrenchLong Jaw
Needle Nose Pliers
PRODUCT NO:

1833

Overall Length: 7" (17,8 cm)
Jaw Length: 2.25" (5,7 cm)
Jaw Width Tapered from: 8 mm to 1.5 mm
Jaw Height Tapered from: 12 mm to 2.5 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY30

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTSSTANDARD LARGEPRODUCT NO’S:

OrthoVise™ Length: 10" (25,4 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3980-01 with Attachment Bolts (two sides & end)
without Slap Hammer
3981
without Attachment Bolts
without Slap Hammer
with End Attachment Nut that accepts aStandard Slap Hammer (#3925 or 3926)

3980

#3980
Side Attachment Bolts

Allows a Large OrthoVise™
Slap Hammer (#3950) to be
attached to the side of the device.

#3980-01

Available on Standard Large and Long Nose
Large OrthoVise™ with Attachment Bolts only.
(Cannot be added on later.)#3981

LONG NOSE LARGEPRODUCT NO’S:

OrthoVise™ Length: 12" (30,5 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3965-01 with Attachment Bolts (two sides & end)
without Slap Hammer

3965

#3965

OrthoVise™

#3965-01

U.S. Patent #D398,208

LONG NOSE LARGE BENT JAW
PRODUCT NO’S:

OrthoVise™ Length: 11.5" (29,2 cm)
with Attachment Nut (end)with Standard Slap Hammer (#3925)
3966-01 without Slap Hammer
with Attachment Nut (end) that accepts aStandard Slap Hammer (#3925 or 3926)

3966

#3966
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY#3966-01

Made of stainless steel
Models equipped with attachment bolts allow aslap hammer to be attached to the end, as wellas to either side of the large OrthoVise™ (except
Bent Jaw models), for greater adaptability
Bent Jaw models are not available with side
attachment bolts, but have an end attachment
nut to accept a Standard
Slap Hammer (#3925 or #3926)
A different size slap hammer is used for thelarge and small sizes of OrthoVise™
Slap Hammers are designed with a hammer
plate for the additionaluse of a mallet if desired

STANDARD SMALLPRODUCT NO’S:

OrthoVise™ Length: 8" (20,3 cm)
without Attachment Bolt
without Slap Hammer
3985-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3985-T with Attachment Bolt (end)without Slap Hammer

#3985

3985

#3985-01#3985-T

LONG NOSE SMALLPRODUCT NO’S:

OrthoVise™ Length: 9.5" (24,1 cm)
without Attachment Bolt
without Slap Hammer
3975-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3975-T with Attachment Bolt (end)without Slap Hammer

3975

SLAP HAMMERS#3975#3975-01#3975-T

PRODUCT NO’S:

3950 [Slap Hammer for Large OrthoVise]For use with 3965's, 3980's, 3981
Overall Length: 16.5" (41,9 cm)

For Large
OrthoViseFor use with: 3975's, 3985's
Overall Length: 8.75" (22,2 cm)

For Small
OrthoViseFor use with: 3966's
Overall Length: 16" (40,7 cm)

Standard
with 16" RodFor use with: 3966's
Overall Length: 16" (40,7 cm)

Easy Grip Standard
with 16" Rod3955 [Slap Hammer for Small OrthoVise]3925 [Standard Slap Hammer w/16" Rod]3926 [Easy Grip Slap Hammer w/16" Rod]

#3950
#3955
#3925
#3926

THREADED ADAPTERS
PRODUCT NO’S:

3980-02 [Small Adapter]#3980-02

Female/Female
Adapter Converts
from Male/MaleChanges Male End of a Slap Hammer to Female

Small Adapter

For use with 3965's, 3966's, 3980's, 3981

Threaded Large#3980-03

For use with: 3975's, 3985's

Threaded Small#3985-03

3980-03 [Threaded Adapting Screw – Large]

3985-03 [Threaded Adapting Screw – Small]

1.800.548.2362       NOVEMBER 2023   

Small Adapter allows a Standard Slap Hammer (#3925 or #3926)
to be used with any Large OrthoVise™ with Attachment Bolts

Threaded Adapting Screws can be used to append the corresponding size
OrthoVise™ with an Attachment Bolt for use with a Slap Hammer   HIP ARTHROPLASTY INSTRUMENTS

31

DMP
Wire Tightener

Four wire holes — twofor up to 20 gauge
wires, and two for up
to 18 gauge wires.
T-Handle end is used
to hand tighten a wire.

Used to hand tighten acerclage wire around a bone
PRODUCT NO:

8729

Overall Length: 4.5" (11,4 cm)
Handle Width: 2.625" (6,7 cm)
End Diameter: 15 mm
Designed by DMP

Incavo Wire Passer
Designed by Stephen J. Incavo, MD

Small #8610-01

Large #8610-02

PRODUCT NO’S:

8610-01 [Small]

Designed to pass multiple cerclage
wires around a bone during a multiple
wire wrap procedure

Overall Length: 7.5" (19,1 cm)
Accepts Wire Up To: 4 mm (5/32")
For Bone Diameter Up To: 1.2" (3 cm)

8610-02 [Large]

Overall Length: 8.675" (21,9 cm)
Accepts Wire Up To: 4 mm (5/32")
For Bone Diameter Up To: 2.4" (6 cm)

Whelan Double-Ended Suture Wire Passer
Designed by Edward J. Whelan, III, MD

Insert passer
into guideto pass aroundthe bone

Passer guide and malleable passer
designed to pass suture wires around a bone

PRODUCT NO’S:

8300-00 [Set with Case]
Also available individually:

8300-01 [Passer Guide]

Overall Length: 8.125" (20,6 cm)
Outside Width: 9 mm
Inside Groove Width: 6,5 mm

8300-02 [Passer]

Overall Length: 7.5" (19,1 cm)
Width: 4,6 mm(one passer with this product number)

1025 [Sterilization Case]

Passer Guide

Set includes passer
guide, two passers, andsterilization case.

#8300-01

Passer #8300-02
Attach suture
wire, then draw
the passer/
suture wire backaround the
bone

32

The passer guide is placed around the bone, andthe thin malleable passer is inserted at the handle end
and follows the grooved passer around. The suture wire (up to 18 gauge)
is attached to the keyholed end of the passer, which can then be reversed out ofthe passer, which can then be reversed out of the passer, drawing the suture wire around the bone.

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

45°

Angled Capsule Scissors
Designed by James B. Stiehl, MD

Angled scissors allow a greater
range of capsular access

20°

PRODUCT NO’S:

45° Capsule Scissors #3079

3079 [45°]Overall Length: 9.5" (24,1 cm)
Scissor Angle: 45°3082 [20°]Overall Length: 10" (25,4 cm)
Scissor Angle: 20°

20° Capsule Scissors #3082

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Malleable Bone Tamp – Extra Small
Modified by Serge Kaska, MD & Amal Das, MD

Designed to help impact bone
into acetabular cup holes
PRODUCT NO:

5296-02 [Extra Small]

Overall Length: 11.4" (29 cm)
Shaft Length: 5.9" (15 cm)
Impactor Diameter: 6.5 mm

Malleable shaft
can be contoured
for different angles

Extended Scalpel Handle

Designed by Richard Pelliccio, MD

Long thin scalpel handle used with knife blade
to make a skin incision and cut throughfascia to help seat trocars to bone
#10 blade normally used but choice of blade isat surgeons’ discretion. Blade not included.
PRODUCT NO:

3022

Overall Length: 18.9" (48 cm)
Handle Length: 5.5" (14 cm)
Shaft Diameter: .25" (6,35 cm)

Laser mark on shaft allows visual orientationof the blade when passing through a cannula.

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

33

Wetzel Acetabular Fragment Clamp
Designed by Robert Wetzel, MD & Todd O. McKinley, MD

Designed to help increase the ability tocontrol and manipulate an acetabular
fragment during Periacetabular Osteotomy(PAO) surgery for hip dysplasia
The cannulated center hinge allows a 5 to 6
mm Schantz pin (not included) to be used inconjuction with the clamp — providing a unified
pin-and-clamp together that is stronger than eachseparately and offers enhanced fragment control.
PRODUCT NO:

3648

Overall Length: 11.5" (29,2 cm)
Jaw Opens to: 1.375" (3,5 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw Width: .5" (12,7 mm)
Hole Diameter for Schantz Pin Up To: .25" (6,3 mm)

Periarticular Reduction Forceps

Designed for reduction of intraarticularand periarticular fractures
Pointed ball tips help provide a secure hold in thebone despite minimal contact. Three sizes available.
PRODUCT NO’S:

1857 [Large]

Large #1857

Medium #1856

Small #1856-01

Jaw Height @ Tips Parallel: 6.25" (15,9 cm)
Jaw Width @ Tips Parallel: 12" (30,5 cm)
Maximum Jaw Opening @ Tips: 8" (20,3 cm)
Overall Length: 16" (40,7 cm)

1856 [Medium]

Jaw Height @ Tips Parallel: 4.75" (12,1 cm)
Jaw Width @ Tips Parallel: 10.5" (26,7 cm)
Maximum Jaw Opening @ Tips: 5.2" (13,2 cm)
Overall Length: 14.75" (37,5 cm)

1856-01 [Small]Jaw Height @ Tips Parallel: 3.375" (8,6 cm)
Jaw Width @ Tips Parallel: 7.25" (18,4 cm)
Maximum Jaw Opening @ Tips: 3.1" (7,9 cm)
Overall Length: 11" (27,95 cm)

Large Bone Clamp
with Plate Protection

Designed to help hold a bone/bone plate in position for
reduction—the one-side coatedjaw helps to protect frommarring the bone plate

Left #3659-L

PRODUCT NO’S:

Right #3659-R

3659-L [Left]

Overall Length: 9.125 (23,2 cm)

3659-R [Right]

Overall Length: 9.125 (23,2 cm)

34

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Subtrochanteric Femur
Fracture Reduction Clamp
Designed by David Beard, MD

Contour design helps clamp asubtrochanteric or femoral shaft
fracture treated with current
generation femoral IM rodding
systems using external aimingarms/targeting devices
PRODUCT NO:

3850

Overall Length: 12.875" (32,7 cm)
Handle Length: 7.5" (19,1)
Jaw Length: 5.25" (13,3 cm)
Jaw Width: .25 (6,3 mm)

Left #3860-L

Cannestra Trochanteric
Fracture Reduction Clamp
Designed by Vince Cannestra, MD

Right #3860-R

Designed to help reduce comminuted
intertrochanteric and subtrochanteric hip
fractures, this clamp is offset at its ends
to avoid placement into the fracture bed
Clamping ends are curved and rotated to allow
maximum bony contact upon fracture reduction. Ideal
for fractures with a flexed anterior cortical spike.
Made for right and left hip fracture configurations.
PRODUCT NO’S:

3860-L [Left]

Overall Length: 11.25" (28,6 cm)

3860-R [Right]

Overall Length: 11.25" (28,6 cm)

Bone Clamp with Speed Lock
Designed to help hold a bone
in position for reduction
PRODUCT NO:

3659

Overall Length: 9.125 (23,2 cm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

35

Browner MIS Bone Clamp
Designed by Bruce D. Browner, MD

Designed to help hold a bone or bone platefor fixation, the clamp is inserted anterior to
the bone, rotated to wrap around the bone,then screwed into the desired position
Sized to allow use on a femur, tibia or humerus.
PRODUCT NO:

1379

Overall Length: 9.25” to 11.5”" (23,5 to 29,2 cm)
Maximum Bone Diameter: ~ 35 mm

Stoll Bone Plate Clamp
Designed by Jordan Stoll, MD
PRODUCT NO:

1774

Overall Length: 10" (25,4 cm)

Designed to help hold a boneor bone plate in position for
reduction and fixation

Chen Diaphyseal Fracture Reduction Clamp
Designed by Franklin Chen, MD

Designed to facilitate and maintain reductionof the internal fixation of diaphyseal andmeta-diaphyseal fractures of long bones
Works especially well with short oblique bones while providing
room to implement the plate with this bone clamp still in place.
Pivoting pads accommodate metaphyseal fractures
The quick release enables adjustment without losing reduction
Helps provide provisional reduction of diaphyseal fractures –humeral shaft fractures, tibial fractures
PRODUCT NO:

1808

Overall Length: 9.25" (23,5 cm)
Arm Downward Offset: 15 mm
Pad Dimensions: 1" x .375" (25,4 cm x 1 cm)

36

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Fracture Reduction Punch Clamp
Designed by Jong-Keon Oh, MD

Designed for use in select cases whenvertical (or sagittal) plane clamping isnecessary during forearm reduction,
humeral fracture reduction, or
diaphyseal reduction of tibial shaft
PRODUCT NO:

5072

Overall Length: 10.5" (26,7 cm)
Point to Point Opening:   –Minimum .375" (10 mm)   –Maximum 1.375" (35 mm)
Pin Diameter: .125" (3,2 mm)

Sumko Surgical Finger Guide

Designed by Michael H. Sumko, MD

Used to help insert a 3.2 mm guide wire, especially during hip
fracture surgery, to help prevent puncturing the surgeons’ glove
The entry point for a trochanteric nail can be located through a smaller incision
with this device, with reduced risk of penetrating the surgeon’s glove whilefinding the starting point for the guide wire.

US Patent #503638945
PRODUCT NO:

8991

Overall Length: 4" (10,2 cm)

Cannulated Fracture Awl

Helps to reduce fractures withoutslipping off the bone, and cannulated
to allow the placement of k-wire

PRODUCT NO:

8091

Overall Length: 7" (17,8 cm)
Handle Length: 3.3" (8,4 cm)
Cannula fits wire up to: .062" (1.6 mm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

37

Soft Impact Mallets

PRODUCT NO’S:

7820 [2 lbs. Standard]with Easy Grip Handles

2 lbs. Standard

Weidman handle designed by Kevin Weidman, MD#7820

Provides shock-absorbing force

Weight: 2 lbs. (.907 kg)
Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7821 [2 lbs. w/Weidman Handle]

Filled with a shock-absorbing media andhas a flat striking surface to keep the mallet
centered on an instrument while providing
less bounce or wasted force.

2 lbs. w/Weidman Handle#7821

The comfortable Easy Grip handle is madeof a textured silicone that helps prevent thesurgeon’s gloved hand from slipping andhelps maintain a solid grip. The bottom canalso be used to tap an implant in place.

2 lbs. w/Delrin End#7832

The mallet with delrin head features a
replaceable delrin head.

Weight: 2 lbs. (.907 kg)
Overall Length: 10.625" (27 cm)
Grip Length: 5.5" (14 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7832 [2 lbs. With Delrin End]
Weight: 2 lbs. (.907 kg)
Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7837 [3 lbs. Standard]

Weight: 3 lbs. (1.35 kg)
Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.875" (4,8 cm)

Delrin Head Replacements for 7832:

3 lbs. Standard#7837

Replacement Delrin Heads

Easy Grip

Comfortable grip helps
prevent the surgeon’s
gloved hand fromslipping and helps
maintain a solid grip.

TexturedSoft Silicone

Handles

Ortho Mallets

Small #7810

PRODUCT NO’S:

7810 [Small]

Overall Length: 8" (20,3 cm)
Handle Length: 4.5" (11,4 cm)
Head Weight: 1 lb. (.45 kg)
Head Diameter: 1.3125"

with Easy Grip Handles
These solid stainless steel mallets
each have a comfortable 4½" grip
made of a textured silicone thathelps prevent the surgeon’s gloved
hand from slipping and helps
maintain a solid grip.

7832-HEAD01 [.5" Stud] Single
7832-HEAD02 [.5" Stud] 3-Pack
7832-HEAD03 [.875" Stud] Single
7832-HEAD04 [.875" Stud] 3-Pack7815 [Large]

Overall Length: 8" (20,3 cm)
Handle Length: 4.5" (11,4 cm)
Head Weight: 1.75 lb. (.8 kg)
Head Diameter: 1.5" (3,8 cm)

Large #7815

Jones Mallet

PRODUCT NO:

Designed by Dickie Jones, MD

7825 [2.4 lbs]

Unique hand fitting shape provides
superior gripping strength

Overall Length: 8.25" (21 cm)
Head Width: 3" (7,6 cm)
Head Diameter: 1.5" (3,8 cm)

This striking instrument has a unique hand
fitting shape that provides superior gripping
strength for accurate light to heavy impaction.

38

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Bechtold Ergonomic
Orthopedic Mallet

Stainless steel head and shaft with an aluminum handle
with a right-handed grip
Large and small striking heads with smooth surface
Palmar side of the mallet features a flat surface to slide
along a broach or impacting type instrument for backslapping and serves well as an additional striking surface

Designed by Dustin Bechtold, MD

Ergonomically designed forforward and backward strikes,
featuring an ergonomic
handle with a tamp
PRODUCT NO:

7822

Overall Length: 10.75" (27,3 cm)
Head Width: 4" (10,2 cm)
Large Head Diameter: 2" (5,1 cm)
Small Head Diameter: 1.5" (3,8 cm)

Aluminum Tapered Maul/Mallet

The large surface area allows the surgeon to focus
on the action area of the instrument being struck,instead of making sure the mallet will strike theend of the instrument, much like a sculptors mallet
PRODUCT NO:

7828 [2.5 lbs]

Overall Length: 9.15" (23,2 cm)
Handle Length: 6" (15,2 cm)
End Diameter: 3" (7,6 mm)

Reusable Light Wand
Designed for illuminationof deep incisions

PRODUCT NO:

8010-02

Can be attached to a fiber optic
light cable with ACMI (female) connector.

Overall Length: 13.5" (34,3 cm)
Handle Length: 5.55" (13,3 cm)
Bend Angle: 45°

Lighted Yankaur Suction Device
Designed by Adolph V. Lombardi Jr., MD

Designed to help provide effective suction
with the addition of a light source forenhanced visualization
Can be attached to a fiber optic light
cable with ACMI (female) connector
Entire device is steam sterilizable
The handle is made of Delrin

Delrin Handle

Light

PRODUCT NO:

8016-L

Suction1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

Overall Length: 11.75" (29,8 cm)
Handle Length: 3.93" (10 cm)
Handle Width: .86" (2,2 cm)
Suction Tube Diameter: .25" (6,35 mm)

39

Ortho Suction Tube

Designed by T. Eickmann, MD

Very effective for suction and minor retracting
Helps eliminate plugging due to bone, cement fragments, blood clots, etc.
PRODUCT NO:

5465

Overall Length: 9.25" (23,5 cm)
End Hole Dia.: 1 mm
Side Hole Dia.: 1.5 mm

Beicker Curette Suction Device

Designed by Clint Beicker, MD

Designed to help visualization of a fracture site
within a fracture hematoma
Also useful for arthroscopic curettage of osteochondral lesions.
PRODUCT NO:

4231

Overall Length: 10.5" (26,7 cm)
Curette Cup: 7.5 mm x 5.5 mm

Blount Retractor with Small Handle
Designed by Ronald Romanelli, MD

A blount retractor with alightweight ergonomic handle
designed for tissue retraction
and closure assistance in knee,shoulder, and hip arthroplasty
PRODUCT NO:

4852

Overall Length: 9.375" (25,1 cm)
Handle Length: 4.625" (11,7 cm)
Blade Depth: 1.5" (3,8 cm)
Blade Width at Widest: .325" (8,3 mm)

Incision Aligner
Designed by DMP

Designed to align an
incision during closing

PRODUCT NO:

1330

Overall Length: 14" (35,6 cm)
Blade Offset: 45 mm

The bent ends of the aligner are placed at each end of an incision, which is aligned by pulling outward on each end. The sliding end
will lock in place when it is tensioned. Pressing inward slightly on the sliding end will allow the aligner to be collapsed and removed.

40

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Gelbke Cobb Elevatorwith Suction
Designed by Martin K. Gelbke, MD

Designed to be used during exposureof the posterior spine, as well as for pelvic
and acetabular trauma cases
Ultra hard titanium nitride coating helps to extend blade life by increasing
surface hardness, prolonging sharpness, and resisting chemicals and corrosion.
PRODUCT NO:

3433

Overall Length: 12.75" (32,4 cm)
Cobb End Width: 18 mm (.7")
Shaft plus Head Length: 5.5" (15 cm)

Cobb Elevators

Two Sizes Available With
or Without Teeth

PRODUCT NO’S:
WITH TEETH

Ultra hard titanium nitride
coating helps to extend blade
life by increasing surface
hardness, prolonging sharpness,and resisting chemicals andcorrosion.

3432 [1/2" with Teeth]

Overall Length: 11" (27,9 cm)
Blade Width: 1/2" (13 mm)

1/2" with Teeth #3432
1/2" without Teeth #3436

3434 [1" with Teeth]

Overall Length: 11" (27,9 cm)
Blade Width: 1" (25,4 mm)

WITHOUT TEETH

3436 [1/2” without Teeth]

Available withor without teeth

Overall Length: 11” (27,9 cm)
Blade Width: 1/2” (13 mm)

3438 [1” without Teeth]1" with Teeth #3434
1" without Teeth #3438

Overall Length: 11” (27,9 cm)
Blade Width: 1” (25,4 mm)

Bradley Periosteal Elevator
Designed by Gary W. Bradley, MD

Ultra hard titanium nitride coating helps to extend
blade life by increasing surface hardness, prolonging
sharpness, and resisting chemicals and corrosion.
PRODUCT NO’S:

4719 [1/2"]

Overall Length: 11" (27,9 cm)
Blade Width: .5" (13 mm)

1/2" #4719

4720 [3/4"]

Overall Length: 11" (27,9 cm)
Blade Width: .75" (19 mm)

3/4" #4720

Periosteal Elevator

PRODUCT NO’S:

3450 [Curved]

Overall Length: 7.5"
Handle Length: 4.5" (11,4 cm)
Blade Size: 16x13 mm

3455 [Straight]

Overall Length: 7.75"
Handle Length: 4.5" (11,4 cm)
Blade Size: 19x14 mm

Designed for better control
Straight #3450

Designed with a curved end for easieruse, and sharper sides for ease ofelevating and stripping. The handle isdesigned for better control.

Curved #3455

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

41

Straight Suture Passer
Designed by Brian T. Maurer, MD

Designed to help pass suture through bone

PRODUCT NO:

1111

Overall Length: 8.125" (20,6 cm)
Handle Length: 4.25" (10,8 cm)
Shaft Diameter: 2,5 mm

The curved semicircular tip is congruent
to most tibial plates and femoral condylar
implants, helping to facilitate removalof excess cement, especially at the tightposterior aspect
The small scoop-end tip assists in excising
unset cement
Ultra hard titanium nitride coating helps
to extend curette life by increasing surface
hardness, prolonging sharpness, andresisting chemicals and corrosion, while
helping to eliminate metal transfer andprotect the implant surface

Sarraf Cement Trimmer
Designed by Khaled M. Sarraf, MD

Two-in-one instrument designed for cement
removal during arthroplasty surgery

PRODUCT NO:

5212

Overall Length: 7.75" (19,7 cm)

The curved semicircular tip is congruent
to most tibial plates and femoral condylar
implants, helping to facilitate removalof excess cement, especially at the tightposterior aspect
The spearhead tip assists in excising andshaping the unset cement
Ultra hard titanium nitride coating helps
to extend curette life by increasing surface
hardness, prolonging sharpness, and resistingchemicals and corrosion, while helping toeliminate metal transfer and protect the
implant surface

Sarraf Spearhead Cement Exciser
Designed by Khaled M. Sarraf, MD

Two-in-one instrument designed for cement
removal during arthroplasty surgery

PRODUCT NO:

5211

Overall Length: 7.75" (19,7 cm)

Cement Packer & Trimmer
Designed by Harlan C. Amstutz, MD
PRODUCT NO:

4995

MADE FOR INNOMED IN

GERMANY

Overall Length: 9.75" (24,8 cm)

42

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

PRODUCT NO:

5635

Overall Length: 8" (20,3 cm)
Freer End: 5 mm
Cup End: 10 mm

Made of Delrin

Robb Cement Curette
Designed by William Robb, MD

Designed to help remove cement
around a hip or knee prosthesis

Straight #5039
Bent #5041

Sarraf TiN Coated
Cement Removal
Forceps
Designed by Khaled M. Sarraf, MD
PRODUCT NO’S:

5039 [Straight]

Overall Length: 6" (15,2 cm)

5041 [Bent]

Overall Length: 6.125” (15,6 cm)

Ultra hard titanium nitride coating helps to extend forceps life by increasing surface hardness, prolonging sharpness,and resisting chemicals and corrosion, while helping to eliminate metal transfer and protect the implant surface.

Bozeman Cement Trimmer
Designed by Daniel M. Gannon, MD

The tool has a blunt blade tip on one end to help withseparation of the trimmed cement. The angled curette end
helps gather the trimmed cement. The thin shank and angled
curette can reach into tight spaces such as the back of theimplants to remove excess cement. The ends are titanium
nitrite coated to help eliminate metal transfer.

Combines the two most common
cement trimming tools into one

PRODUCT NO:

5245

Overall Length: 8.5" (21,6 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Seachris Delrin Cement Scraper
Designed by Timothy Seachris

Reusable delrin scraper is designed
to help remove cement around aknee or hip prosthesis
PRODUCT NO:

5218

Overall Length: 5" (12,7 cm)
Thickness: 1/8" (3.1 mm)

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

43

Bates Needle Holder with Suture Cutter

No switching between needle
driver and scissors, or need forassistant to cut sutures for you
Cutting with opening of forceps
reduces possibility of damage tosurrounding tissues
Sliding the instrument down to
the suture knot allows quick andconsistent 2 mm suture tails
Slip the suture strands into
the suture cutting slot andslide the closed instrument
along until desired length oftail is achieved, then open the
instrument to cut the sutures

Designed by James E. Bates, MD

Suture Cutter
Needle Holder

By trapping the suture and cutting when theforcep is opened, helps to reduce stresson the surgeon’s hand

PRODUCT NO:

3071

Overall Length: 8.125" (20,6 cm)
Jaw Width: .25" (6,4 mm)
Open Jaw Length: .5" (12,8 mm)

Stanton Needle Driver

Allows a heavy cutting needle
such as an OS-6 to be pushed
through cancellous bone when
re-attaching muscle or tendon

Designed by John L. Stanton, MD, FACS

The groove captures the outer (convex) sideof the needle and prevents the needle fromspinning even when applying significant
pressure. Useful for reattaching the rotator
cuff in rotator cuff repairs, as well as in
attaching suture anchors.

PRODUCT NO:

3042

Overall Length: 6.75 (17,1 cm)
Jaw Width: .25" (6,3 mm)

Long Bonney
Tissue Forceps

Extra length—3" more thanstandard—allows for use in deep wound areas
PRODUCT NO:

5040

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 10" (25,4 cm)

GERMANY

Charnley Type Tissue Needle Forceps

Helpful for wound closure in deep areas with fascia
under tension such as hip or knee replacement
Can also help retrieve a needle in a tight area.
Designed by Amal Das Jr., MD

PRODUCT NO:
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

441165

Overall Length: 6.875" (17,5 cm)

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Wilke Angled
Blunt Nose Scissors
Designed by Benjamin K. Wilke, MD

Allows blunt dissecting
around critical structures(nerves, vessels, etc.)
while maintaining a cutting
surface for fascia. The
tool's blunt ends can also
be used for cauterizing andgrabbing small vessels.

PRODUCT NO:

3078

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 6" (15,2 cm)

GERMANY

Orthopedic Needle
Holder/Scissors
Drive a needle andcut a suture withoutchanging instruments
5.5" Tungsten
Carbide #3055

6.5" Tungsten
Carbide #3065

7" Tungsten
Carbide #3075

Scissors
Holding
Tips
MADE FOR INNOMED IN

GERMANYPRODUCT NO’S:
Standard Tips

7" Standard

3070 7.0" (17,8 cm)#3070

Tungsten Carbide Tips

3055 5.5" (14 cm)
3065 6.5" (16,5 cm)
3075 7.0" (17,8 cm)

Longer sizes arehelpful in orthopedics

Rogozinski Locking Needle Driver/Scissors

Designed with a quick lock & release handle, can drivea needle and cut a suture without changing instruments
Standard #3083

Scissors
Holding Tips

Large #3084
PRODUCT NO’S:

3083 [Standard]

Designed by
Chaim Rogozinski, MD

3084 [Large]

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 6.5" (16,5 cm)

Overall Length: 7.75" (19,7 cm)

GERMANY

Sweed Dissecting Scissors
Designed by Tamer Sweed, FRCS (Orth)

Designed with a blunt, flat bar
fixed to the lower limb, the scissors
also act as a dissector to protect
underlying vital structures
PRODUCT NO:

3081

Overall Length: 6.625" (16,8 cm)
Bottom Pad: 16 mm x 6 mm)
Pad Extension Beyond Scissor: 6 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

45

Large T-Handle
Fixed Drivers

Large easy grip soft silicone
handled drivers help provide
a sturdy non-slip grip
The two standard Quick-connect
models release by pulling thecollar backward, while the Reverse
Quick-connect model is designed
to have the collar be pushedforward for release.

Zimmer Hall
Quick-connect

HudsonQuick-connect

PRODUCT NO’S:

8248 [Zimmer Hall Quick-connect]
Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

Zimmer Hall
Quick-connect#8248

8248-01 [Reverse Quick-connect Zimmer Hall]

Zimmer Hall
Reverse Quick-connect

Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

8249 [Hudson Quick-connect]

HudsonQuick-connect#8248-01

Overall Length: 6.75" (17,1 cm)
Overall Length with Pin In Handle: 11.5" (29,2 cm)#8249

T-Handle Chuck & Key
PRODUCT NO’S:

8247-00 [T-Handle Chuck & Key]8247-01 [T-Handle Chuck Only]
8247-02 [Chuck Key Only]

Gray Syringe Assist with Ergonomic Handle

Designed by Robert Gray, MD

For use in the O.R or the office, the design helps to prevent hand fatigue
and pain when injecting with a 20mL syringe over multiple cases
Sterilizable for O.R use, such as injecting the posterior capsule during TKA
Especially useful for injecting preoperative local anesthesia for WALANT surgery
Uses finger flexors to generate more force over more surface area than only the thumb flexor
Ratchet mechanism ensures maximal grip force generation throughout entire injection

Syringe not included.

PRODUCT NO:

8988

Overall Length – Closed: 5.25" (13,3 cm)
Overall Length – Open: 7.5" (19,1 cm)
Height: 5" (12,7 cm)
Syring Diameter: 21 mm

Patent Pending

White Aspiration Handle

Designed by Edward White, MD

Designed for aspiration of cavities or spacesthat have greater than 20 ml volume, suchas joints, bone marrow, and the illiac crest
Works with a 60 ml syringe (not included) only.
Syringe not included.

PRODUCT NO:

1131

Height: 5" (12,7 cm)
Length: 6.5" (16,5 cm) / Extends to 11" (27,9 cm)
Width at Syring Holder: 1.5" (3,8 cm)
Body Width: .9" (2,3 cm)

46

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Universal Screwdriver Set
Set consists of one handle
and one sterilization/storage
case, plus seven double
ended screwdriver bits:

Helps eliminate the openingof multiple sterile packs whena specific size or style ofscrewdriver is needed

Handle#5195-01

Helpful during revision total joint surgery wherescrews have been used, removal of bone plates,
fracture fixation screws or bone graft screws.
PRODUCT NO’S:

Small & Large Single Slot #5195-02

5195 [Complete Set with Case]

Cross & Cruciate #5195-03

5195-01 [Handle]
5195-02 [Straight (single slot)]

Also sold individually

Large: 7 x 1.5 mm, Small: 5 x 1 mm

3.5 mm & 4.5 mm Hex #5195-04

5195-03 [Cross/Cruciate]

Small & Large Phillips #5195-05

5195-04 [Hex]Large: 7 mm, Small: 6 mm

Large: 4.5 mm, Small: 3.5 mm

5195-05 [Phillips]#6 & #8 star #5195-08

Large: 4 mm, Small: 3.5 mm

5195-08 [Small Star: #6 & #8]
5195-06 [Medium Star: #10 & #15]
5195-07 [Large Star: #20 & #25]

#10 & #15 star #5195-06#20 & #25 star #5195-07

Star Bit Driver Set
Helps eliminate theopening of multiple
sterile packs when aspecific size of star
bit is needed

T25

PRODUCT NO’S:

5194-00 [4 Star Bits w/Handle & Case]
5194-01 [4 Star Bits w/Case only]

T20

Set in Storage Case

#5194-25#5194-20

Also sold individually:

T15

S0113 [Universal 4" (10,2 cm) Handle]
5194-10 [T10 with A/O End]5194-15 [T15 with A/O End]5194-20 [T20 with A/O End]5194-25 [T25 with A/O End]
9003 [Case]#5194-15

T10#5194-10

Torx/Hex Adapter Set

PRODUCT NO’S:

8003-00 [Set – One Each]

Designed by Stephen M. Walsh, MD

Set Includes/Available Separately:

Designed for conversion
of a 3.5 mm screwdriver

8003-01 [Torx Bit to Hex Driver Adapter]
Overall Length: .6" (1,54 cm)

8003-02 [Hex Bit to Torx Driver Adapter]
Overall Length: .6" (1,54 cm)

Helpful during revision total joint surgery. Set
consists of four star bits — T10, T15, T20, &T25, a handle which accommodates any of
the above bits, and a sterilization case. The
drive end (A/O) is designed for easy and quick
engagement with the universal instrument
handle. The ergonomic, modular handle has twoconnection points, allowing for both straightand T-handle orientations.

Especially helpful when an articulated, universal
joint driver is needed (i.e. acetabular screws)

Hex Bit to Torx Driver Adapter#8003-02

Torx Bit to Hex Driver Adapter#8003-01

Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

1.800.548.2362       NOVEMBER 2023      HIP ARTHROPLASTY INSTRUMENTS

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:

47

PRSRT. STD.
U.S. POSTAGE
PAID
GRAND RAPIDS, MI

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 748

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

Beicker Hammerhead Rongeur
Designed by Clint Beicker, MD

Designed to help remove osteophytes fromaround the acetabulum and glenoid
PRODUCT NO:

1775-05

Overall Length: 8" (20,3 cm)
Jaw Bite: 15 mm x 7 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYINNOMED, INC

Innomed-Europe LLC

TOLL FREE 1.800.548.2362

Tel 912.236.0000
Fax 912.236.7766

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74

Innomed-Europe GmbHISO 13485:2016

103 Estus Drive
Savannah, GA 31404

Villingen-Schwenningen,Deutschland
Tel 0049 (0) 7720 46110 60

innomed.net
info@innomed.net© 2023 Innomed, Inc.
All Rights Reserved

11
23

www.innomed-europe.com
orders@innomed-europe.com

INNOMED.NET

info@innomed.net