J A NMUA AY R Y

2023

OR T HOPEDIC INS T RUMEN T S

Vosburg
Cannulated Periarticular Clamp
Page 7®

Malleable
Bone Tamps
Page 19

Chandran
Double Ball SpikePage 22
Fromm
Femur & Tibia Triangles
Page 3Large T-Handle
Fixed
Drivers
Page 44

Stoll
Retractor/Wire Guide
Page 28

Trauma Instruments
1.800.548.2362 • INNOMED.NET

Soft Impact Mallets

PRODUCT NO’S:

7820 [2 lbs. Standard]

Weight: 2 lbs. (.907 kg)
Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

with Easy Grip Handles
Weidman handle designed by Kevin Weidman, MD

Provides shock-absorbing force

7821 [2 lbs. w/Weidman Handle]
Weight: 2 lbs. (.907 kg)
Overall Length: 10.625" (27 cm)
Grip Length: 5.5" (14 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

Filled with a shock-absorbing media and has a flat
striking surface to keep the mallet centered on an
instrument while providing less bounce or wasted force.
The comfortable Easy Grip handle is made of a textured
silicone that helps prevent the surgeon’s gloved handfrom slipping and helps maintain a solid grip. The
bottom can also be used to tap an implant in place.
The mallet with delrin head features a replaceable
delrin head.

7832 [2 lbs. With Delrin End]
Weight: 2 lbs. (.907 kg)
Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7837 [3 lbs. Standard]

Weight: 3 lbs. (1.35 kg)
Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.875" (4,8 cm)

Delrin Head Replacements for 7832:

7832-HEAD01 [.5" Stud] Single
7832-HEAD02 [.5" Stud] 3-Pack
7832-HEAD03 [.875" Stud] Single
7832-HEAD04 [.875" Stud] 3-Pack

Easy Grip Handles
Textured Soft Silicone
Comfortable grip helps
prevent the surgeon’s
gloved hand fromslipping and helps
maintain a solid grip.

Replacement Delrin Heads

PRODUCT NO’S:

7810 [Small]

Ortho Mallets

Overall Length: 8" (20,3 cm)
Handle Length: 4.5" (11,4 cm)
Head Weight: 1 lb. (.45 kg)
Head Diameter: 1.3125"

with Easy Grip Handles

7815 [Large]

These solid stainless steel mallets each have a
comfortable 4½" grip made of a textured siliconethat helps prevent the surgeon’s gloved hand fromslipping and helps maintain a solid grip.

Jones Mallet

Overall Length: 8" (20,3 cm)
Handle Length: 4.5" (11,4 cm)
Head Weight: 1.75 lb. (.8 kg)
Head Diameter: 1.5" (3,8 cm)

Designed by Dickie Jones, MD

PRODUCT NO:

Unique hand fitting shape provides
superior gripping strength7825 [2.4 lbs]

Overall Length: 8.25" (21 cm)
Head Width: 3" (7,6 cm)
Head Diameter: 1.5" (3,8 cm)

This striking instrument has a unique hand fitting shapethat provides superior gripping strength for accurate
light to heavy impaction.

Aluminum Tapered Maul/Mallet
The large surface area allows thesurgeon to focus on the action areaof the instrument being struck,instead of making sure the mallet
will strike the end of the instrument,much like a sculptors mallet

2

PRODUCT NO:

7828 [2.5 lbs]

Overall Length: 9.15" (23,2 cm)
Handle Length: 6" (15,2 cm)
End Diameter: 3" (7,6 mm)

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

By twisting the dial,
the device can beraised, lowered or
rotated as needed

Jackson Flat Top Traction Device
A table-top traction device designed for fracture
fixation in the acetabulum, pelvis, and femur
Can be used in a variety of applications, including open andpercutaneous pelvic and acetabular fracture surgery, hip fracture
fixation and femur fracture fixation including antegrade or
retrograde nailing.
The light-weight portable device attaches directly
to a standard radiolucent flat top table. Recommended
for use with the disposable sterile kit, one of which isincluded, with additional kits sold separately.
PRODUCT NO’S:

0007 [Jackson Traction Device]

This product number includes (1) #0008 Disposable Sterile Kit

Sold Separately:

0008 [Disposable Sterile Kit]

Kit Includes: (1) Impervious Stockinette and (1) 11 ft. Traction rope
0008-CASE [Case of Disposable Sterile Kits] Pkg of 10

Shown with
Disposable
Sterile Kit

Mounts tothe base of
the OR table(weight not included)

Fromm Femur & Tibia Triangles
Used for femur and tibia positioning
during nailing, repairs and fractures
Designed by S.E. Fromm, MD *Extra Small Triangle designed by S.E. Fromm, MD & Kenneth Merriman, MD

Designed to position and hold the femur and tibia
during intramedullary nailing of the tibia, ligament
repairs and extremity fractures. Allows knee to be flexed
greater than 90° to allow reaming and nail insertion
without displacing fracture. The triangles are available infour heights: 8.5", 11", 14", and 16". The three smaller
triangles are designed to fit inside the larger triangle forstorage. They are supplied with an autoclavable silicone
cushioning pad and velcro* straps. The triangles arealso radiolucent and gas or steam sterilizable.

PRODUCT NO’S:

2760-00 [Set of 3] Angles: Top 30°, Two Bottom 75°
2760-01 [11"] Base: 6" (15,2 cm), Height: 11" (27,9 cm)
2760-02 [14"] Base: 7" (17,8 cm), Height: 14" (35,6 cm)
2760-03 [16"] Base: 9" (22,9 cm), Height: 16" (40,7 cm)
Sold Separately – Not In Set:

2760-XS [8.5"] Base 5" (12,7 cm), Height: 8.5" (21,6 cm)
Replacement Parts:

Tibia Reduced For:
• Open Reduction and Internal Fixation (ORIF)• Application of uni- or multi-plane external fixator• Knee ligament repairs and/or reconstruction

RetrogradeFemoral Nailing

RetrogradeFemoral Nailing

Tibial Nailing

Triangle holds femur reduced(prevents sagging)

1.800.548.2362       MAY 2023   

2760-P [Silicone Pad]
2760-S [Straps] Package of 18
8120-SP [Straps for XS] Package of 10*Velcro® is a registered trademark of the Velcro Companies.

   TRAUMA INSTRUMENTS

3

Auerbach Arm Holder Rake Retractor Set
Designed by David M. Auerbach, MD

Allows intraoperative positioning for proceduresof the posterior arm, elbow, and forearm

PRODUCT NO’S:

2415-00 [Auerbach Arm Holder Rake Retractor Set]
Individual/Replacement Parts:

2415-01 [Auerbach Arm Holder Assembly]

Simple design for fast andeasy positioning
Connects over the drape in thesterile field using the supplied
rail clamp and post
Can be repositioned during
surgery
Sterilizable rubber pad
protects the arm
Retractors for the skin and soft
tissues connect to the holder
Compact for easy storage

Overall Length: 20" (50,4 cm)
Arm Holder Dimensions: 14.5" x 4" (36,9 x 10,2 cm)
Overall Width inclusing Cleats: 7.5" (19,1 cm)

2415-02 [Auerbach Arm Holder Upright Rod]Overall Length: 19.25" (49,9 cm)

2415-04 [Auerbach Rake Chain Retractor 4-Prong]
Two included in set, one with this product number
Overall Length including Chain: 10" (25,4 cm)
Retractor Width: .75" (1,9 cm)

2415-06 [Auerbach Rake Chain Retractor 6-Prong]
Two included in set, one with this product number
Overall Length including Chain: 10" (25,4 cm)
Retractor Width: 1.25" (3,2 cm)

2590-S [Black Strap – Pkg of 10]

Two straps included in set, ten with this product number
Dimensions: 1" x 24" (2,5 x 61 cm)

2595 [Table Clamp]
2770-P [Silicone Pad]

Set includes: Arm Holder Upright and Assembly, (2) 4-Prong
Chain Retractors, (2) 6-Prong Chain Retractors, (2) Black
Straps, Silicone Pad, and Table Clamp.

Dimensions: 12" x 5.5" (30,5 x 14 cm)

Distal Humerus Fracture Board
Designed by Burk Young, MD

Designed for the pinning of pediatric supracondylar and adult distal humerus fractures
Allows the surgeon to pin these fractures without having to manually
hold the fracture reduced, allowing the surgeon to focus on accurate pin
placement and reduction. The height of the crossbar is fully adustable toaccommodate different size patients. Reduction is acheved by an assistantgently applying axial traction through the forearm, with the crossbar
applying the counter traction. Pinning is done with the C-arm in the lateralposition. An optional separate attachment to support the arm for distal
humerus fractures in adults is available. Unit not sterilizable.

Board, crossbar, andoptional adult adapter
are radiolucent.

PRODUCT NO’S:

2445 [Fracture Board – Pediatric]

Main Board Dimensions: 22" x 12" (55,8 cm x 30,5 cm)
Crossbar Height Adjusts From: 4.5" to 7.5" (11,4 cm x 19,1 cm)

2445-01 [Fracture Board – With Adult Adapter]
Optional/Replacement Part:2445-06 [Adult Adapter]

4

Optional

ADULT

Adapter

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Adjustable Knee & Tibial Positioner
Designed by Ashutosh Chaudhari, MD

Adjustable design allows for use in procedures around theknee such as tibial nailing, tibial condyle plating, patella
fracture fixation, supracondylar fracture plating, supracondylar
fracture nailing,and total knee replacement
PRODUCT NO’S:

2770-00 [Set]

Includes Positioner, Pad,
and Two Short Straps

Individual / Replacement Parts

2770-01 [Positioner]

Overall Length (Folded): 28" (71,1 cm)
Overall Length (Flat): 54.75" (139 cm)
Maximum Triangle Height: 14" (35,6 cm)
Width: 5.5" (14 cm)
Thickness (Folded): 1.8" (4,6 cm)
Thickness (Flat): .75" (1,9 cm)

2770-P [Slicone Pad]

Dimensions: 12" x 5.5" (30,5 cm X 14 cm)

2590-S [Short Straps] Pkg of 10

Radiolucent. Steam sterilizable.

Sanders Extremity Positioning Tubes

Designed by
Richard A. Sanders, MD

Designed to support the knee and ankle duringlower extremity surgery
The 6" tube lifts the knee off the operating table and allows for approximately 30°
of knee flexion. Very useful for closure of total knee incisions, supporting fracturesof the distal femur, and tibia plateau fractures. The 4" tube elevates the foot andankle for ankle fracture surgery. The tubes are made of aluminum, allowing them
to be autoclaved. They help eliminate the need for rolled sheet bolsters.
PRODUCT NO’S:

2740-01 [Small]

Diameter: 4" (10,2 cm)
Width: 8" (20,3 cm)

2740-02 [Large]

Diameter: 6" (15,2 cm)
Width: 8" (20,3 cm)

Lower Extremity Leg Positioner
Designed by Ronald Romanelli, MD

Designed to lift the knee for lower
extremity casting applications
Also well suited for use with ankle fractures.
Supplied with one autoclavable silicone pad.
Positioner is radiolucent and gas or steam sterilizable.
PRODUCT NO’S:

2745

Dimensions: 5.5" H x 9.5" L x 9.25" W      (12,7 cm x 24,1 cm x 23,5 cm)

Replacement Parts:

2760-P [Silicone Pad]

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

5to help reduce long bone fractures of the femur and
AK Fracture Reducer Designed
tibia, especially helpful with shortened long bone fractures due
Designed by Byron McCord, MD

to young, strong musculature in acute trauma, or neglected
fractures due to overriding circumstances or late referral

PRODUCT NO:

Serrated
Tip

3870

Overall Length: 12.5" (31,8 cm)
Blade Width: 1.5" (3,8 cm)

The curved, serrated tip helps to wedge and holdthe reducer in place
The curved trough side of the reducer helps capture
and control the bone while leverage is applied
Once in place, by pushing on the T-handle, thesurgeon uses the reducer to help move the bones
into alignment for plating or rodding

Initialengagement

Initial activation

Completion of
distraction/reduction
and disengagement
TroughFemur/Tibia Fracture Distractor

Use with most bone clamps for overlapped
diaphyseal fractures (fig. 1) or 6 mm Schanz
pins to distract intra-articular fractures(fig. 2) for reduction and fixation

Bone clamps and Schanz pins not included.

Fig. 1: Diaphyseal fractures with clamps

Fig. 2: Intra-articular fracture using pins

PRODUCT NO’S:

1809

Overall Length: 10.5" (26,7 cm)
Overall Width: 7.25" (18,4 cm)
For Pins Up To: .25" (6,4 mm)
Individual/Replacement Parts:

1809-02 [Pivot Block]
1809-03 [Frame (no pivot blocks or moveable arm)]
1809-04 [Moveable Arm (no pivot block or handle)]
1809-05 [Handle]
A portion of all proceeds goes to SIGN Fracture
Care International, a 501(c)(3) non-profit, topromote equality of fracture care in developing
countries. signfracturecare.org

6

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Periarticular
Reduction Forceps

Pointed ball tips
help provide a secure hold in
the bone despite minimal contact

Designed for reductionof intra-articular andperiarticular fractures
PRODUCT NO’S:

1856-01 [Small]Jaw Height @ Tips Parallel: 3.375" (8,6 cm)
Jaw Width @ Tips Parallel: 7.25" (18,4 cm)
Maximum Jaw Opening @ Tips: 3.1" (7,9 cm)
Overall Length: 11" (27,95 cm)

1856 [Medium]

Jaw Height @ Tips Parallel: 4.75" (12,1 cm)
Jaw Width @ Tips Parallel: 10.5" (26,7 cm)
Maximum Jaw Opening @ Tips: 5.2" (13,2 cm)
Overall Length: 14.75" (37,5 cm)

1857 [Large]

Jaw Height @ Tips Parallel: 6.25" (15,9 cm)
Jaw Width @ Tips Parallel: 12" (30,5 cm)
Maximum Jaw Opening @ Tips: 8" (20,3 cm)
Overall Length: 16" (40,7 cm)

Vosburg Cannulated Periarticular Clamp
Designed by Caleb Vosburg, MD

Cannulated clamp tips allow passage of k-wires
By compressing the fracture with the clamp and then passing
two k-wires, the clamp can then be removed to allow moreworking room and versatility when applying a plate.

PRODUCT NO:

1864

Overall Length: 13" (33 cm)
Handle Length: 8" (20,3 cm)
Ratcheted Opening from 2" to 3.5" (5,4 to 7,6 cm)
Accepts Pins up to: 7/64" (2.8 mm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Ratcheting Reduction Clamp Assembly
Designed by Michael Craig, OPA-C

Designed as a soft tissue sparing
fracture reduction clamp
PRODUCT NO’S:

3840-00 [Clamp Assembly]
Also available Individually:

3840-02 [Plate Point]

Overall Length: 1" (2,54 cm)

3840-03 [Screw Point]

Overall Length: .875" (2,2 cm)

3840-04 [Percutaneous Point]

2 included in assembly, one with this product number
Overall Length: 1" (2,54 cm)

Plate
Point

3840-MA [Ratcheting Reduction
Mobile Arm with Ratchet Knob]
Overall Length: 6.5" (16,5 cm)

3840-SA [Ratcheting Reduction
Stationary Arm]
Overall Length: 10.5" (26,7 cm)
Width: 9" (22,9 cm)
Height: 6" (15,2 cm)

Screw
Point

Percutaneus
PointsAssembly includes:(1) Ratcheting Reduction Stationary Arm,
(1) Ratcheting Reduction Mobile Arm with Ratchet Knob,(1) Plate Point, (1) Screw Point, and (2) Percutaneous Points

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

High torque can help provide bone and joint
reduction without squeezing surrounding tissues
Swivel points are placed on the bone, plate, or
screw and the ratcheting dial is turned to thedesired torque, allowing hands free operation
Swivel point design allows the clamp to be easily
moved from x-ray view without losing reduction
Screw Point fits into a screw head
Plate Point fits into a 3.5 mm plate hole7

Browner MIS Bone Clamp

Sized to allow use on a femur, tibia or humerus.

Designed by Bruce D. Browner, MD

Designed to help hold a bone orbone plate for fixation, the clamp isinserted anterior to the bone, rotatedto wrap around the bone, thenscrewed into the desired position
PRODUCT NO:

1379

Overall Length: 9.25” to 11.5”" (23,5 to 29,2 cm)
Maximum Bone Diameter: ~ 35 mm

Durkan Ratchet Bone Clamps

Designed by John Durkan, MD

Design of ratcheting mechanism
allows for quick tightening andrelease around the bone
PRODUCT NO’S:

1867 [Large]

Overall Length: 8.625" (21,9 cm)
Jaw opens to: 3.5" (8,9 cm)

1868 [Small]

Overall Length: 8.5" (21,6 cm)
Jaw opens to: 3.75" (9,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Large Bone Clamp
with Plate Protection

Designed to help hold a bone/boneplate in position for reduction—the
one-side coated jaw helps to protectfrom marring the bone plate
PRODUCT NO’S:

3659-L [Left]

Overall Length: 9.125 (23,2 cm)

3659-R [Right]

Overall Length: 9.125 (23,2 cm)

8

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Cannestra Trochanteric
Fracture Reduction Clamp

Designed to help reduce
comminuted intertrochanteric andsubtrochanteric hip fractures, thisclamp is offset at its ends to avoid
placement into the fracture bed
Clamping ends are curved and rotated toallow maximum bony contact upon fracture
reduction. Ideal for fractures with a flexed
anterior cortical spike. Made for right andleft hip fracture configurations.
PRODUCT NO’S:

3860-L [Left]

Overall Length: 11.25" (28,6 cm)

3860-R [Right]

Overall Length: 11.25" (28,6 cm)

Subtrochanteric Femur Fracture Reduction Clamp
Designed by David Beard, MD
PRODUCT NO:

3850

Overall Length: 12.875" (32,7 cm)
Handle Length: 7.5" (19,1)
Jaw Length: 5.25" (13,3 cm)
Jaw Width: .25 (6,3 mm)Contour design helps clamp a subtrochanteric or
femoral shaft fracture treated with current generation femoral
IM rodding systems using external aiming arms/targeting devices

Bone Clamp with Speed Lock
Designed to help hold a bone
in position for reduction
PRODUCT NO:

3659

Overall Length: 9.125 (23,2 cm)

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

9Redler Wrist Bone Clamp
with Wire Guide

PRODUCT NO’S:

1885-45

For Pins up to .045" (1,1 mm)
Overall Length: 9.5" (24,1 cm)
Jaw opens to: 3.5" (8,9 cm)

Designed by M.R. Redler, MD

1885-62

For Pins up to .062" (1,6 mm)
Overall Length: 9.5" (24,1 cm)
Jaw opens to: 3.5" (8,9 cm)

Designed to hold bony
fragments in place forplacement of guide wires
Can be used for:Placement of pins acrossdistal radius fractures or
across carpal bones
Arthroscopically assisted
fixation in the wrist
Fracture fragments about
the elbow
Placement of guide wires
during the open reduction
and internal fixation of apatella fracture

Two sizes available:
For use with 0.045" (1.1 mm)
or 0.062" (1.6 mm) K-wires.

Wetzel Acetabular Fragment Clamp
Designed by Robert Wetzel, MD & Todd O. McKinley, MD

Designed to help increase the ability tocontrol and manipulate an acetabular
fragment during Periacetabular Osteotomy(PAO) surgery for hip dysplasia
The cannulated center hinge allows a 5 to 6 mm
Schantz pin (not included) to be used in conjuction
with the clamp — providing a unified pin-and-clamp
together that is stronger than each separately andoffers enhanced fragment control.
PRODUCT NO:

3648

Overall Length: 11.5" (29,2 cm)
Jaw Opens to: 1.375" (3,5 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw Width: .5" (12,7 mm)
Hole Diameter for Schantz Pin Up To: .25" (6,3 mm)

Pivoting pads accommodate
metaphyseal fractures
The quick release enables
adjustment without losing
reduction
Helps provide provisional
reduction of diaphyseal
fractures – humeral shaft
fractures, tibial fractures

Chen Diaphyseal Fracture Reduction Clamp

Designed by Franklin Chen, MD

Designed to facilitate and maintain reductionof the internal fixation of diaphyseal andmeta-diaphyseal fractures of long bones
Works especially well with short oblique bones
while providing room to implement the platewith this bone clamp still in place.

PRODUCT NO:

1808

Overall Length: 9.25" (23,5 cm)
Arm Downward Offset: 15 mm
Pad Dimensions: 1" x .375" (25,4 cm x 1 cm)

10

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Durham Bone Reduction Clamps
Designed by Alfred A. Durham, MD

Allows application of a bone plate
without removing the reduction clamp

The standard clamp is designed
for medium size bones such as
the fibula, ulna, and radius

The large window directly above the jaws provide
space to allow a bone plate to be slid into
position without removing the clamp.

Standard

PRODUCT NO’S:

3652 [Standard]

Overall Length: 7.375" (18,7 cm)

3652-01 [Large with Speedlock]
Overall Length: 9.25" (32,5 cm)

Large Window

Provides space to allow a boneplate to be slid into position
without removing the clamp

The large clamp with speedlock
is designed for large bones suchas the femur and tibia

Wixted Fracture Distactor

Large with
Speedlock

Designed by John J. Wixted, MD

Designed to provide opposing leverage to help bring
the fibula (or other bone) back out to its proper
length after it has been shortened by a fracture

PRODUCT NO:

1882

Overall Length: 7" (17,8 cm)

A 3.5 mm screw is temporarily placed above a plate,providing a source of leverage for the screw holding end of
the distractor. The curved peg-shaped tip is then placed intoa hole in the bone plate, and the distracter is activated tobring the bone back to its proper length before fixation.
Cut-out for Screw

Curved Peg-shaped Tip

Fits securely into a hole in abone plate for leverage

Provides a secure source of leverageagainst a temporarily placed 3.5 mm screw

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

11

Chen Low Profile Plate/Bone Clamp
Designed by Franklin Chen, MD

The rotating nut is used totighten/release the center arm
once in the desired position

Designed for plate to bone
clamping in a variety of lower
and upper extremity fractures

The freely swivelingcenter arm allows foreasy placement, as well
as for quick release, aftergetting the legs in position

Useful for diaphyseal forearm fractures,
humerus fractures, and distal radius fractures.
PRODUCT NO:

1639

Overall Length: 4.5" (11,4 cm)
Prong Depth: 1" (2,5 cm)
End Prong Width: 6 mm|16 mm Gap|6 mm
Clamps from: .375" to 2.4" (1 to 6 cm)

Stoll Bone Plate Clamp
Designed by Jordan Stoll, MD

Designed to help hold a bone orbone plate in position for reduction
and fixation—helpful with clavicle
and fibula fractures

PRODUCT NO:

1774

Overall Length: 10" (25,4 cm)

Fracture Reduction Punch Clamp
Designed by Jong-Keon Oh, MD

Designed for use in select cases whenvertical (or sagittal) plane clamping isnecessary during forearm reduction,
humeral fracture reduction, or
diaphyseal reduction of the tibial shaft
PRODUCT NO:

5072

Overall Length: 10.5" (26,7 cm)
Point to Point Opening:   –Minimum .375" (10 mm)   –Maximum 1.375" (35 mm)
Pin Diameter: .125" (3,2 mm)

12

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Calvo Fracture Reduction Clamp
Designed by Ignacio J. Calvo, MD
PRODUCT NO’S:

1801-L [Left]
1801-R [Right]

Medial Malleolus
Fracture Reduction

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Designed to reduce and hold a displaced
medial malleolus fracture, as well as
to reduce and hold in place transverse
fractures of the olecranon, to facilitate theinsertion of k-wires and tension bands

Lateral Condyle Fracture Set

Olecranon
Fracture
Reduction

The stops on each clamp end help
prevent excessive penetration of
metaphyseal and soft bone

Designed by Carl R. Weinert, MD

Designed for adult and pediatric lateral condyle fractures

Symmetric Reduction Clamp

Useful to compress T-condylar fractures, andin many other fracture reduction applications.
The stops on each end help prevent excessive
penetration of metaphyseal and soft bone.

PRODUCT NO'S:

4697-00 [Set with Case]
Set Includes / Available Individually:

1755 [Clamp – Symmetric]

Asymmetric Reduction Clamps – Left and Right

Shaped to secure the lateral condyle fragment. The straight tip isplaced in the coronoid fossa and the curved tip is used to graspand compress the lateral condyle fragment. The stops on each end
help prevent excessive penetration of metaphyseal and soft bone.

Overall Length: 8.5" (21,6 cm)
Jaw opens to: 3" (7,6 cm)

1756-L [Clamp – Asymmetric Left]
Overall Length: 8.75" (22,2 cm)

1756-R [Clamp – Asymmetric Right]

Elbow Retractor

Overall Length: 8.75" (22,2 cm)

Designed for use within the elbow joint to
retract the anterior capsule, and provide full
exposure of the anterior articular surface for
reduction and fixation of displaced lateralcondyle fractures. The small blunt ti
hooks over the intact medial condyle.

4697 [Elbow Retractor]

Overall Length: 6.5" (16,5 cm)
Blade Width: 1" (2,54 cm)

1015 [Sterilization Case]

Dimensions: 11.25" x 7.125" x 3.125" (28,6 cm x 18,1 cm x 7,9 cm)

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

13

Pointed Fracture Reduction Clamps
Designed by Reza Firoozabadi, MD MA

Versatile set of fracture reduction clamps, each witha specific tine design that allows for appropriate
vector placement so that anatomic reduction can beobtained in a number of different types of fractures

Straight Left & Right

1.9 mm tines allow for a snug fit in 2 mm drill holes
Tines angled to prevent clamp “slippage” with compression
Straight tines can be placed deep within bone which allows for far cortex compression.
Clamps incorporate a box joint design that prevents clamp joint loosening and theneed for tightening.
Example applications: any transverse fracture (straight-straight clamp), both bone
forearm fractures, olecranon fractures, medial malleolus fractures, and many more.
Speed Lock Style: Extra-long spin down allows for increased range of clamp use, andopen-topped joint rotates to allow for increased range of opening, and also allows forquick release
PRODUCT NO’S:
SMALL WITH SPEED LOCK MECHANISM

Small with
Speed Lock

3666-01 [Straight Left & Right]

3667 [Curved Left & Right]

3667-01 [Curved Left & Right]

Overall Length: 5.5" (14 cm)

Small with
RatchetMedium with
Speed Lock

MEDIUM WITH SPEED LOCK MECHANISM

3666 [Straight Left & Right]
Overall Length: 5.5" (14 cm)

Curved Left, Straight Right

Overall Length: 7" (17,8 cm)
Overall Length: 7" (17,8 cm)

3666-L [Curved Left, Straight Right] 3666-L-01 [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

Overall Length: 7" (17,8 cm)

Overall Length: 5.5" (14 cm)

Overall Length: 7" (17,8 cm)

3666-R [Straight Left, Curved Right] 3666-R-01 [Straight Left, Curved Right]

Two styles — Speed Lock and Ratchet — each avalable in four tine configurations

SMALL WITH RATCHET MECHANISM

3668 [Straight Left & Right]
Overall Length: 5.5" (14 cm)

3669 [Curved Left & Right]
Overall Length: 5.5" (14 cm)

3668-L [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

3668-R [Straight Left, Curved Right]
Overall Length: 5.5" (14 cm)

Straight
Left & Right

CurvedLeft & Right

Curved Left,
Straight Right

Straight Left,
Curved RightO’Brien Bone Clamps
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
Allows for placement of the bone clamp where
it can best stabilize bone fragments. The drill
guide allows for screw placement through the
top of the clamp. Calibrations on the handle
help eliminate the use of a depth gauge.

PRODUCT NO’S:

1890-02 [Large]

Drill Guide Diameter: 10 mm(accomodates up to 6.5 mm screw)
Calibrated from 12 mm to 40 mm
Overall Length: 9.25" (23,5 cm)

Integrated drill
guide and bonediameter gauge

10 mm Drill Guide Diameter(accommodates up to 6.5 mm screw)

1890-01 [Small]

Drill Guide Diameter: 8 mm
(accomodates up to 4 mm screw)
Calibrated from 8 mm to 30 mm
Overall Length: 6" (15,2 cm)

8 mm Drill Guide Diameter(accommodates up to 4.0 mm screw)

1890-XSM* [Extra Small]
Drill Guide Diameter: 6 mm
Overall Length: 4” (10,2 cm)

MADE EXCLUSIVELYFOR INNOMED IN*

GERMANY

6 mm Drill Guide Diameter

14

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Stanton Articulating
Small Bone Clamps
Designed by John L. Stanton, MD

Compress with a towel clamp
Distract with a gelpi retractor

Opposing clamps
facilitate manipulationof fracture ends

PRODUCT NO’S:

1811-00 [Set of Two]
Also available individually:

The small tube allows use of
a towel clamp to compress
non-union and shortening
osteotomies during fixation, as
well as to allow the use of Gelpi
retractors to distract malunions
during revision surgery.

1811-L [Left]

Overall Length: 5.125" (13 cm)
Curved Plate Radius: 5 mm
Pin Hole for Pins Up To: 2,4 mm

1811-R [Right]

Overall Length: 5.125" (13 cm)
Curved Plate Radius: 5 mm
Pin Hole for Pins Up To: 2,4 mm

Medial Malleolar/Bone Fragment Clamps
Designed by Edward L. Sclamberg, MD

Quick tightening & release low profile
clamp with unlimited settings
PRODUCT NO’S:

1830 [Standard]

Overall Length: 5.5" (14 cm)
Clamp End Length: 1"

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY1835 [Medium]

Overall Length: 6" (15,2 cm)
Clamp End Length: 2"

1840 [Large]

Overall Length: 8" (20,3 cm)
Clamp End Length: 3"

Teurlings Medial Malleolar
Clamp with Wire Guide

Designed by Luc Teurlings, MD

Helps to stabilize the medial malleolar
fragment during internal fixation
MADE EXCLUSIVELYFOR INNOMED IN

GERMANYPRODUCT NO:

1803

Cannula Diameter: .062" (1.6 mm)
Overall Length: 5.25" (13,3 cm)

Bargo Bone Holding Clamp
Designed by Lonnie Bargo, CST/CFA

Designed to aid in the reduction of various fractures,and can help secure a plate in place during installation

PRODUCT NO:

1895-01

Overall Length: 5" (12,7 cm)
Pads: .75" x .45" (19 x 12 mm)

Designed to aid in the reduction of various fractures such as:
spiral, transverse, compound, oblique, or butterfly. The clamp
can also be used to secure a plate in place while the screw
holes are being drilled and screws inserted. The fracture site
can also be manipulated with the clamp being used as alever. Teeth in the jaws allows for a better grip and a ratchet
locking handle allows use on various bone diameters.

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

15

Faillace Extra Small Bone Clamp
Designed by John J. Faillace, MD

Delicate enough to use on metacarpals but
strong enough for distal radius and larger
bones with its extra long ratchet
PRODUCT NO:

1171

Overall Length: 5" (12,7 cm)
Jaw Length: 1" (2,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYO’Brien Bone Clamp
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
PRODUCT NO:

1816

Overall Length: 5.25" (13,3 cm)

OrthoLucent O’Brien
Bone Clamp
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
The carbon fiber PEEK material is strong, lightweight,
completely radiolucent, can be steam sterilized, andhelps to prevent from marring component surfaces.
PRODUCT NO:

1815-R

Overall Length: 5.25" (13,3 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

Small Bone Holding Forceps
with Long Ratchet
Designed for use in stabilizationof a fracture or osteotomy
PRODUCT NO:

1170

Overall Length: 5.75" (14,6 cm)

16

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Redler Percutaneous Pin Clamp

Holds a small bone in apposition duringpercutaneous pinning of a fracture
Designed with a proximal pin tube with teeth; the tube guides
the pin and the teeth help keep the tube in place on the bone.
The distal tip is used to control the bone fragment. Includes along ratchet for locking on various sized bones, from 1 mm to 14
mm. Also useful during insertion of cannulated screw guide wires.

Three Tube Sizes
Available

PRODUCT NO’S:

Overall Length: 5" (12,7 cm)

1810-35 Tube Diameter: .035" (.9 mm)
1810-45* Tube Diameter: .045" (1.1 mm)
1810-62 Tube Diameter: .062" (1.6 mm)

MADE EXCLUSIVELYFOR INNOMED IN*

GERMANY
Designed by M.R. Redler, MD

Chang Pin Clamp
Designed by Win Chang, MD

Designed to allow accurate insertionof pins for internal fixation
Used for small bones, the clamp allows
accurate insertion of pins for internal fixation.
The cannula has a 1.8 mm internal diameter.
PRODUCT NO:

1760-01(Prototype used in
X-ray images)

Cannula Internal Diameter: 1.8 mm
Overall Length: 6" (15,2 cm)
Locking Ratchet Opens To: 25 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Slavitt Phalangeal Forceps
Designed by Jerome Slavitt, DPM

Designed to enable the surgeon
to provide joint distraction andstability during joint placement at
the base of the proximal phalanxof the lesser digits
PRODUCT NO:

Helps to distract the joint and hold the bone,allowing easier access to the base. Can also be
used for digital fusions to hold bones better for
drilling and cutting applications.

MADE EXCLUSIVELYFOR INNOMED IN

1163

Overall Length: 6" (15,2 cm)
Clamp Internal Opening Diameter: 4 mm

GERMANY

Lewin Small Bone Clamp
PRODUCT NO:

4685

Overall Length: 5" (12,7 cm)

1.800.548.2362       MAY 2023   

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   TRAUMA INSTRUMENTS

17

Coated Allis Bone Clamps

Modification of design by Charles T. Resnick MD

A traditional Allis Bone Clamp designed with a longer ratchet—for a wider
opening to allow a bone and plate to be clamped and locked onto—andcoated end(s) to prevent from marring a component surface
PRODUCT NO’S:

1381 [One Coated End]

Overall Length: 6.125" (15,9 cm)
Ratcheted Clamp Opens to: 35 mm
Non-coated-end Width: 4 mm

1382 [Two Coated Ends]

Overall Length: 6.125" (15,9 cm)
Ratcheted Clamp Opens to: 35 mm
Non-coated-end Width: 4 mm

Resnick Allis Bone Clamp
Designed by Charles T. Resnick MD

A traditional Allis Bone Clamp
designed with a longer ratchet whichallows for a wider opening to allow abone to be clamped and locked onto
PRODUCT NO:

MADE EXCLUSIVELYFOR INNOMED IN

1385

Overall Length: 6" (15,2 cm)
Ratcheted Clamp Opens to: 37 mm
Clamp End Width: 4.7 mm

GERMANY

Standard hooked bone awl being used to gain length
to assist with reduction of a fibula fracture. A 2mm
pilot hole is made to seat the tip of the bone awl.

Hooked Bone Awls
Designed by Reza Firoozabadi, MD

PRODUCT NO’S:

5078 [Standard]“Shoulder hook” awls designed to help withmanipulation of bone fragments for fixation

Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)

5078-01 [Long]

Overall Length: 13.375" (34 cm)
Handle Length: 6" (15,2 cm)

Fracture Reduction Pick

Used to align bone fragments, and topick away tissue and bone fragments
PRODUCT NO:

S0129

Overall Length: 6.25" (15,9 cm)

18

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Malleable Bone Tamps

Modified by Serge Kaska, MD

The large tamp is designed to help elevate a depressed tibial plateau
fracture, while the small tamp can help elevate a depressedtibial plafond and smaller tibial plateau fractures
PRODUCT NO’S:

5296 [Large]

Overall Length: 14" (35,6 cm)
Shaft Length: 9.5" (24,1 cm)
Impactor Diameter: 12.5 mm

5296-01 [Small]

Overall Length: 9.5" (24,1 cm)
Shaft Length: 6" (15,2 cm)
Impactor Diameter: 10 mm

5296-02 [Extra Small]

Overall Length: 11.4" (29 cm)
Shaft Length: 5.9" (15 cm)
Impactor Diameter: 6.5 mm

Malleable shaft can be contoured for different angles

Large

12.5 mm

Small

10 mm

Extra Small
6.5 mmBacastow Tibial Plateau Elevators
Designed by David Bacastow, MD

Designed to help with indirect
reduction of a depressed tibial
plateau fracture, and canbe used with arthropscopic
visualization andpercutaneous fixation

Finish10.4 mm

PRODUCT NO'S:

5297 [Starter Elevator]

Starter

Overall Length: 11" (27,9 cm)
Tamp Diameter: 4,7 mm

4.7 mm5298 [Finish Elevator]

Overall Length: 11" (27,9 cm)
Tamp Diameter:10,4 mm

Sandman Curved Bone Punch
Designed by Geoffrey A. Sandman, MD

Designed to help elevate a depressedtibial plateau fracture
PRODUCT NO:

5305

Overall Length: 14" (35,6 cm)
Shaft Length: 9.5" (24,1 cm)
Impactor Diameter: 12,5 mm (.5")

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

19

Resnick Small Bone Tamp with Oblique K-Wire Hole
Designed by Charles Resnick, MD

Design allows for the concurrent reduction ofa fracture and placement of a wire into thefracture site — especially helpful when
the surgical exposure is small andtight, the fracture fragments are
small, and the reduction isdemanding
TWO SIZES AVAILABLE:
Wire Hole for K-wires up
to 1.1 mm (.045”) or
1.6 mm (.062”)

BeforeThe serrated distal end minimizes slippage on the cortical surface,does not interfere with the placement of the guidewire and allows for
subsequent surgeon-decided, intraoperative angulation of the wiring
once the first cortex is drilled
Especially useful in fractures where there is involvement of an articularsurface, for example, mallet fractures of the distal phalanx, articular
fractures that involve ligamentous attachments or tendon attachmentsof the phalanges, scaphoid pole small fracture fragments or other small
carpal fractures, and radial styloid fractures

After

Extended Drill Sleeves

PRODUCT NO’S:

5294 [1.2 mm Hole]

Wire Hole for: 1,2 mm (.045") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

5294-01 [1.6 mm Hole]

Wire Hole for: 1,6 mm (.062") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

Designed by Reza Firoozabadi, MD

Designed to help reduce fractures when K-wires are passed
through, the extra long drill sleeve helps to protect soft tissues
and prevent the need for stacking two drill sleeves

2.4/1.8 mm

2.7/2.0 mm

Serrated tips allow for better grip
when drilling at an angle or whenpushing a fracture fragment toassist with fracture reduction
Sleeve can be used as a reduction
aid with placement of a kirschner
wire through sleeve
Collaborated tips which allow
placement of appropriate size drills
for lagging by technique — as an
example a 2.5 end will fit into a
3.5 drill hole
Case example of using modified
3.5/2.5 mm drill sleeve for
placing 3.5 mm screws in a
forearm fracture case. Note
how extended sleeve protects
soft tissues during drilling.

3.5/2.5 mm
PRODUCT NO’S:

3014-00 [Set of Three]
Set Includes/ Available Separately:

3014-01 [2.4/1.8 mm]

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

ProfileExtra Long Guide Sleeve Length: 2.25" (5,7 cm)

Larger Diameter

Serrated Sleeve Ends

20

Collaborated tip end diameters
allow placement of appropriate size drills
for lagging by technique — as an example a
2.5 end will fit into a 3.5 drill hole.

3014-02 [2.7/2.0 mm]
Smaller Diameter

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

3014-03 [3.5/2.5 mm]
Serrated Sleeve Ends

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

WWW.INNOMED.NET   

Note 2.0 mm end of drill sleeve
placed into a predrilled 2.7
mm hole, utilized as a lag bytechnique 2.7 mm screw.

   FREE TRIAL ON MOST INSTRUMENTS

Zell Fixed Angle Wire Guide
Designed by Richard Zell, MD

Designed to help with placement ofguide wires for cannulated screws
and k-wires in foot and ankle
surgeries, such as bunion surgery,midfoot fusion, and midfoot ORIFPRODUCT NO:

3021

Overall Length: 2.75" (7 cm)
Handle Platform: 1" x .875" (25 x 22 mm)
Guide Tube Angle: 35°
Guide Tube for wires up to: .052"/1.3 mm

Medial Malleolus
Fracture Reduction Aid
Designed by Christopher Blair, DO

Designed to hook under the medial malleolus
to help reduce the medial malleolus fragment while
two K-wire guides supply trajectory for wires
For K-wires up to 1,6 mm (.062")
PRODUCT NO:

3664

Overall Length: 7" (17,8 cm)
Handle Length: 4" (10,2 cm)
Plate Width: .8" (2 cm)
Plate Length: 3" (7,6 cm)
Guide Tube Length: 6 mm

Mogul K-Wire/Pin Insertion Guide
Designed by Stuart J. Mogul, DPM, FACFAS

A guide designed for passing guidepins or k-wires through two adjacentmetatarsal bones
PRODUCT NO:

3017

Dimensions: 2.375" Tall x 3.75" Wide (6 x 9,5 cm)
Maximum Pin Diameter: 3/32" (2,4 mm)
Maximum Clamped Opening: 2" (5,1 cm)
Minimum Clamped Opening: .375" (1 cm)
Pin/K-Wire Guide Length: .925" (23,5 mm)

Argintar Claw Drill Guide Wire/Suture Passer

PRODUCT NO’S:

Designed by Evan Argintar MD

8315-00 [Set: (1) Claw,(1) Wire/Suture Pin]
8315-01 [Claw Unit]

Expandable claw design allows for minimally invasive,
reproducible one-step wire/suture passage

Overall Dimensions: 2.5" x 4"–6"(6,4 cm x 10,2 cm–15,2 cm)

Especially helpful during applications where a suture will bepassed—particularly when soft tissue dissection is to beminimized, such as wrist reconstruction (DRUJ), elbow
reconstruction (ULCL/MCL), foot-ankle reconstruction(ATFL), quad/patella tendon repair surgery, andmulti-ligament knee reconstruction (MCL/LCL).

1.800.548.2362       MAY 2023   

1227 [3/32" (2 mm) Pin
with Wire/Suture Hole]
Overall Length: 6" (15,2 cm)   TRAUMA INSTRUMENTS

21

Chandran
Double Ball Spike
Designed by Rama E. Chandran, MD

Designed to help rotate and control
a butterfly bone fragment for fixation
PRODUCT NO:

8027

Overall Length: 12.5" (31.8 cm)
Handle Length: 4.625" (1,17 cm)

Ball Spike with Bell Handle

Designed with a long shaft for use in deep wounds
PRODUCT NO:

8032

Overall Length: 12" (30,5 cm)

Cannulated Fracture Awl

Helps to reduce fractures withoutslipping off the bone, and cannulated
to allow the placement of k-wire

PRODUCT NO:

Cannulated

8091

Overall Length: 7" (17,8 cm)
Handle Length: 3.3" (8,4 cm)
Cannula fits wire up to: .062" (1.6 mm)

Small Cannulated Ball Spike
Designed by Benjamin C. Taylor, MD

Designed to help reduce a bone
fragment and keep it reduced, while thecannulation allows placement of a k-wire(up to 1.6 mm/.062") into the fragment
PRODUCT NO:

Helps to prevent slipping while inserting k-wires
Can serve as a handle for k-wire joysticks

22

8092

Overall Length: 4.5" (11,4 cm)
Handle Length: 3" (7,6 cm)
Ball Diameter: .275" (7 mm)

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Roberts Pin Bending Cannula Set

Features

Designed by David Roberts, MD

Helpful when removing a pin
which has been cut flush to thebone which can be hard to grasp
with standard tools.
PRODUCT NO’S:

2113-00 [Set of 3 Sizes]
Set Includes / Available Individually:

2113-01 [2 mm]

Designed to help bend the end of a flexible
intramedullary pin, which has been cut flush to thebone, for better grasping during pin removal

Beveled edge that allows
easy capture of end of pin
Cannula can be used as a
trephine for pins with bonyovergrowth
Reusable - thick walls
withstand repeated uses

After exposing the pin end, the cannula helps bend
the pin for better access for the removal
instrument while maintaininga small incision.
4 mm

For 1.5 - 2.0 mm flexible nails
Overall Length: 5.5" (14 cm)

3 mm

2113-02 [3 mm]

2 mm

For 2.5 - 3.0 mm flexible nails
Overall Length: 5.5" (14 cm)

2113-03 [4 mm]

For 3.5 - 4.0 mm flexible nails
Overall Length: 5.5" (14 cm)

Technique

Distal Radius – 1.5 cm Incision

Proximal Ulna – 1.5 cm Incision

Advance cannula over the exposed pin.
The beveled end makes this easier andcan be used as a trephine in when bonyovergrowth is present.
Slowly use the tool to bend theend of the pin away from bone, whichhelps with pin removal.

The bent end can be removed
with standard pliers.

Stanton Nail/Screw Drill Guide Assembly
FOR DISTAL HUMERAL, FEMORAL, OR TIBIAL SCREWS
Designed by John L. Stanton, MD

Designed to help hold and stabilize a drill guide, allowing the surgeon
to obtain 'perfect circles' and drill distal locking screw holes
without exposure of the hand to the x-ray beam
The drill guide is placed over the side of the bone through an incision. The holder
is attached to the guide and rested against the skin for stability. With the x-ray
on, the guide tube is adjusted by moving the holder until the guide lines up
with the hole in the rod. A drill bit is then inserted into the guide.

PRODUCT NO:

8986-00 [Assembly Set]

Set includes: (1) Holder, (1) Sleeve, and (1) Trocar
Also available individually:

8986-01 [Drill Guide Sleeve]
Overall Length: 3.85" (9,8 cm)
Outside Diameter: 7 mm

Locking Drill Guide Holder

8986-02 [Drill Guide Trocar Alignment Tool]
Overall Length: 4.375" (11,1 cm)
Trocar Diameter: 5 mm

Trocar Alignment Tool

8987 [Locking Drill Guide Holder]
Overall Length: 10.5 (26,7 cm)
Guide Height: 21 mm

Sleeve1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

23

Incavo Wire Passer
Designed by Stephen J. Incavo, MD

PRODUCT NO’S:

8610-01 [Small]

Overall Length: 7.5" (19,1 cm)
Accepts Wire Up To: 4 mm (5/32")
For Bone Diameter Up To: 1.2" (3 cm)

Designed to pass multiple cerclage
wires around a bone during a multiple
wire wrap procedure

8610-02 [Large]

Overall Length: 8.675" (21,9 cm)
Accepts Wire Up To: 4 mm (5/32")
For Bone Diameter Up To: 2.4" (6 cm)

DMP Wire Tightener
Designed by DMPUsed to hand tighten acerclage wire around a bone
PRODUCT NO:

8729

Four wire holes — two for up
to 20 gauge wires, and twofor up to 18 gauge wires.

Overall Length: 4.5" (11,4 cm)
Handle Width: 2.625" (6,7 cm)
End Diameter: 15 mm

Beard IM Nail Guide Wire Clamp

Designed to help provide quick graspand-release of an IM guide wire forpositioning and advancement along
the length of the guide wire

Designed by David Beard, MD

Anatomic pistol grip for comfortable use
Facilitates fracture reduction in appropriate cases
Universal to all fixation systems using IM guidewires and IM fixation
For use with pins up to 4 mm

Available with orwithout ratchet

PRODUCT NO’S:

3019 [Beard Clamp with Ratchet]

Dimensions: 5.5" w x 6" h (14 cm x 15,2 cm)

3019-01 [Beard Clamp without Ratchet]
Dimensions: 5.5" w x 6" h (14 cm x 15,2 cm)

Sumko SurgicalFinger Guide
Designed by Michael H. Sumko, MD

Used to help insert a 3.2 mm guidewire, especially during hip fracture surgery,to help prevent puncturing the surgeons’ glove
PRODUCT NO:

8991

Overall Length: 4" (10,2 cm)

24

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

K-Wire Bender/Cutter
1 and
1.2 mm

1.2 and
1.6 mmThe right slot of the
instrument’s lower
jaw can hold K-Wireswith a diameter of
1.2 mm or 1.6 mm.
The smaller left slot
can hold K-Wires
measuring 1 mm or
1.2 mm in diameter.

Bending

Smooth Bending
Clean Cutting

Designed to bend a K-Wire while extending
from bone without applying mechanical strain
The K-Wire only needs to extend 20 mmfrom the skin surface to be bent.

Can bend and cut K-Wires
measuring 1 to 1.6 mm(.039–.062") in diameter
PRODUCT NO:

2111

Overall Length: 6.5" (16,5 cm)

With the jaw of the instrument
opened wide, the K-Wire isinserted from the side into oneof the slots of thelower jaw. During bending, theK-Wire is forced backwards by thenose of the upper jaw and guided
by a small groove.

Cutting

The K-Wire is inserted into
the cutting groove and thebender/cutter cuts by shearing(like a cigar cutter), notcrushing. The result is a clean
and burr-free cut surface.

Pin Puller - Small

Small size allows for use
in a small incision to help
with removal of a 2 mm or
smaller k-wire pin
PRODUCT NO:

3033

Overall Length: 6.5" (16,5 cm)
Jaw Width: 6,2 mm tapering to 3 mm at end
Jaw Height: 11,7 mm

Stanton Bent Pin Removal Pliers
Designed by John Stanton, MD, FACS

For pins up to .062" (1,6 mm).
PRODUCT NO:

1894

Overall Length: 6.5" (16,5 cm)
Jaw Length: 1.65" (4,2 cm)
Instrument Width: 1 cm

Stanton Straight Pin Removal Pliers
Designed by John Stanton, MD
PRODUCT NO:

1893

Overall Length: 6.375" (16,2 cm)
Jaw Length: 1.62 (4,1 cm)
Instrument Width: 1 cm

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

25

Lighted Yankaur Suction Device
Designed by Adolph V. Lombardi Jr., MD

Designed to help provide effective suction
with the addition of a light source forenhanced visualization
Can be attached to a fiber optic light
cable with ACMI (female) connector
Entire device is steam sterilizable
The handle is made of Delrin

Delrin Handle
Light

PRODUCT NO:

8016-L

Overall Length: 11.75" (29,8 cm)
Handle Length: 3.93" (10 cm)
Handle Width: .86" (2,2 cm)
Suction Tube Diameter: .25" (6,35 mm)

SuctionWhelan Double-Ended Suture Wire Passer

Designed by E. J. Whelan, III, MD

Passer guide and malleable passer designed to pass suture wires around a bone
Insert passer
into guideto pass aroundthe boneSet includes Passer Guide, twoPassers, and a sterilization case.
PRODUCT NO’S:

8300-00 [Set]
Also available individually:

8300-01 [Passer Guide]

Overall Length: 8.125" (20,6 cm)
Oustide Width: 9 mm
Inside Groove Width: 6.5 mm

8300-02 [Passer]   Two included with set,   one with this product number
Overall Length: 7.5" (19,1 cm)
Width: 4.6 mm

1025 [Sterilization Case]

The passer guide is placed around the bone, andthe thin malleable passer is inserted at the handle end
and follows the grooved passer around. The suture wire (up to 18 gauge)
is attached to the keyholed end of the passer, which can then be reversed out ofthe passer, which can then be reversed out of the passer, drawing the suture wire around the bone.

Attach suture
wire, then draw
the passer/
suture wireback aroundthe bone

Straight Suture Passer
Designed by Brian T. Maurer, MD

Designed to help pass suture through bone
PRODUCT NO:

1111

Overall Length: 8.125" (20,6 cm)
Handle Length: 4.25" (10,8 cm)
Shaft Diameter: 2,5 mm

26

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Gain Radiolucence
Without Compromising
Strength or Function

OrthoLucent™ Retractors
PRODUCT NO’S:

2820-R [PCL]

Overall Length: 9.875" (25,1 cm)
Prong Width: 5 mm | 10 mm Gap | 5 mm

Ideal for total joint use with x-ray
Steam sterilizable
Completely radiolucent
Flat black non-gloss finish
No metal transfer with component contact
Slightly higher cost than stainless steel

3220-02R [Chandler]

Overall Length: 9.875" (25,1 cm)
Blade Width: 19 mm

PCL
Retractor

6130-R [Cobra]

Overall Length: 12" (30,5 cm)
Blade Width at Widest: 33 mm#2820-R

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

Chandler
Retractor#3220-02R

OrthoLucent™
Hohmann Retractors

Cobra Retractor

PRODUCT NO’S:#6130-R4535-R [Modified Narrow]
Blade Width: 18 mm
Overall Length: 10"

Modified Narrow Hohmann

4550-R [Modified Blunt]#4535-R

Blade Width at Widest: 24.5 mm
Overall Length: 10.75"

4558-R [Standard]

Blade Width: 16 mm
Overall Length: 9.625" (24,4 cm)

Modified Blunt Hohmann#4550-R

7110-R [Narrow Bent]

Overall Length: 9.75" (23,8 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 19 mm
Depth from Bend: 4.75" (12,1 cm)

Standard Hohmann#4558-R

MADE EXCLUSIVELYFOR INNOMED IN

Narrow Bent Hohmann

SWITZERLAND#7110-R

Carbon fiber PEEK
composite material is strong,lightweight, completely
radiolucent and can be
steam sterilized, and helps
to prevent from marring
component surfaces

Modified
Hohmann Retractors
Handle is contoured toallow better leverageand visualization
Useful for retracting tissues
around the bone. Can beheld in place with weights
or by hand.
Narrow#4535

OrthoLucent™ Narrow#4535-R

PRODUCT NO’S:

4535 [Narrow]

Overall Length: 10" (25,4 cm)
Blade Width: 14 mm

4535-R* [OrthoLucent™ Narrow]
Overall Length: 10" (25,4 cm)
Blade Width: 18 mm

Short-tipped Narrow

4535-01 [Extra Deep Narrow]#4545

Overall Length: 11.625” (29,5 cm)
Blade Width: 16.4 mm

Extra Deep Narrow#4535-014545 [Short-tipped Narrow]
Designed by Carl DiRaimondo, MD

WideOverall Length: 9.5" (24,1 cm)
Blade Width: 14 mm#6595

6595 [Wide]

Overall Length: 10" (25,4 cm)
Blade Width: 42.5 mm

6595-01 [Extra Deep Wide]
Overall Length: 11.5" (29,2 cm)
Blade Width: 42.5 mm*OrthoLucent™ The OrthoLucent™ version is made of a strong, lightweight
MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

carbon fiber PEEK composite material, which is completely
radiolucent, helps to prevent from marring component
surfaces, and can be steam sterilized.

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

Extra Deep Wide
#6595-0127

Stoll Retractor/Wire Guide

Designed by Jordan Stoll, MD

Retractor/guide designed for aiming guide wire when performing
femoral nailing (TFN for an intertrochanteric fracture), or
tibial nailing using the parapatellar approach
PRODUCT NO:

8012

Overall Length: 12" 30,5 cm)
Blade Width: .71" (1,8 cm)
Prong Length: 6 mm
Guide Hole Diameter: 4,5 mm

McPherson Retractor Extender
Designed by Ed McPherson, MD

Designed to extend a standard retractor
to help provide additional leverage
Available in two sizes to accomodate most retractors— standard for retractors up to .125" (3,2 mm) thick,and large for retractors up to .16" (4 mm) thick.

PRODUCT NO'S:

6022 [Standard]

Overall Lentgth: 15.625" (39,7 cm)
For retractors up to .125” (3,2 mm) thick6022-01 [Large]

Overall Length: 15.625" (39,7 cm)
For retractors up to .16” (4 mm) thick

Stainless Steel Hip Surgery Ratchet Frame with Arms and Blades Sets
Designed for self-retaining wound exposure, the arms and bladesof the OrthoLucent™ version are radiolucent and can be kept inplace while using image intensification or taking an x-ray
Arms rotate 180°
Mobile Arm unit can be detachedfrom ratchet body for cleaning

Set with OrthoLucent™
Arms and Blades

The OrthoLucent arms and blades are made of a strong,lightweight carbon fiber PEEK composite material, whichis radiolucent, helps to prevent from marring component
surfaces, and can be steam sterilized.

Set with
Stainless SteelArms and Blades

OrthoLucent Parts
MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

PRODUCT NO’S:

7428-00 [Stainless Steel Ratchet Frame with
OrthoLucent Arms & Blades Set]

Stainless
Steel Parts

PRODUCT NO’S:

7429-00 [Stainless Steel Ratchet Frame with
Stainless Steel Arms & Blades Set]

Set Includes / Available Individually:

7428-01 [Stainless Steel Ratchet Frame
with OrthoLucent Arms Assembly]
Dimensions (Flat): 10" x 5.625" (25,4 cm x 14,3 cm)
Arms Extend: 4.25" (10,8 cm)

7427-02 [50 mm OrthoLucent Blade]Dimensions: 50 mm Deep X 25 mm Wide7427-03 [75 mm OrthoLucent Blade]

Set Includes / Available Individually:(1) 50 mm & (1) 75 mm blade
included in each set.
Optional 100 mm blade
available separately.

Dimensions: 75 mm Deep X 25 mm WideDimensions (Flat): 10" x 6" (25,4 cm x 15,3 cm)
Arms Extend: 4.875" (12,4 cm)

7429-02 [50 mm Stainless Steel Blade]Dimensions: 50 mm Deep X 25 mm Wide7429-03 [75 mm Stainless Steel Blade]Dimensions: 75 mm Deep X 25 mm Wide

Optional Blade – Not Included In Set:

Optional Blade – Not Included In Set:

7427-04 [100 mm OrthoLucent Blade]7429-04 [100 mm Stainless Steel Blade]Dimensions: 100 mm Deep X 25 mm Wide287429-01 [Stainless Steel Ratchet Frame
with Stainless Steel Arms Assembly]

Dimensions: 100 mm Deep X 25 mm Wide

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Dozier Radiolucent
Bennett Hip
Fracture Retractor
Designed by John K. Dozier, MD

Handle canbe rotated toright or left

Designed to be used in hip fractures with theadvantage that the retractor can be kept in place
while using image intensification or taking an x-ray.
The handle can be rotated to the right or left forsurgeon preference. May be steam or gas sterilized.

Can be kept in place while
using image intensification
or taking an x-ray
PRODUCT NO:

6870

Handle Length: 6.75" (17,1 cm)
Blade Length: 8.5" (21,6 cm)
Blade Width at Widest: 67 mm

Kaminsky OrthoLucent™ Browne-type Deltoid Retractors
Used for the Delto-Pectoral Approach—can remainin place for fracture reduction, plate positioning,and screw/wire/drill location confirmation

Kaminsky
Retractor

Completely radiolucent

Used for acromioplasty, rotator cuff repair, and fracture fixation. Contours thehumeral head with deltoid retraction allowing extensive exposure. Helps to reduce
operative time, assist in fracture reduction, and maintain hardware position without thefrequent need for retractor removal and reintroduction.
Carbon fiber PEI composite material is strong, lightweight, completely radiolucent,can be steam sterilized, and helps to prevent from marring component surfaces.
PRODUCT NO’S:

1670-01R [Small]

Blade Width: 4,5 cm
Overall Length: 10.5" (26,7 cm)

Designed by Sean B. Kaminsky, MD

1670-02R [Large]

Blade Width: 5,4 cm
Overall Length: 10.5" (26,7 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

Levy Wide Deltoid Retractor
Designed by Jonathan Levy, MD

Designed for management of proximal
humerus fractures—facilitates appropriatedeltoid retraction without interference
during active fluoroscopy
Patent Pending
PRODUCT NO:

1672

Overall Length: 11.75" (29,8 cm)
Blade at Widest: 2.5" (6,4 cm)
Blade Depth: 1.375" (3,5 cm)

Contoured to match the curve of the deltoid, the retractor
helps to retract the entire deltoid laterally during thedeltopectoral approach. The width approximates 2/3 the lengthof the deltoid, while the blade is deep enough to help control theentire deltoid without displacement of the tuberosity reduction.
Sized to fit deltoids in small and large patients.

Taylor Retractors
PRODUCT NO’S:

6330-01 [Standard]

Overall Length: 8" (20,3 cm)
Depth from Bend: 4" (10,2 cm)
Blade Width: 32 mm

6330-02 [Large]

Overall Length: 9" (23 cm)
Depth from Bend: 5.5" (14 cm)
Blade Width: 32 mm

6330-03 [Deep with Pin Guides]
Overall Length: 9" (23 cm)
Depth from Bend: 5.5" (14 cm)
Blade Width: 32 mm
Guide for Pins Up To: .125" (3,2 mm)

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

29

Rosen “V” Deep Soft Tissue Retractor
Designed by Adam Rosen, DO

Designed for soft tissue retraction with an ergonomic handle
PRODUCT NO:

6239

Overall Length: 12" (30,5 cm)
Blade Depth: (3.5" (8,9 cm)
Blade Width: 1.75" (4,4 cm)

Wide Rake Retractors
with Ergonomic Handle
Designed for general use
soft tissue retraction,
the ergonomic handle
allows for a better grip
and less fatigue
Non-glare finish featured on
the metal retractor parts.

Deep,
Blunt

PRODUCT NO’S:

6051 [Deep, Sharp]

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 2.75" (7 cm)

6052 [Deep, Blunt]

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 2.75" (7 cm)

6053 [Shallow, Sharp]

Sharp and dull teeth
available in two depths

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 1.875" (4,8 cm)

6054 [Shallow, Blunt]

Shallow,
Sharp

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 1.875" (4,8 cm)

PRODUCT NO’S:

Meyerding Type Retractors
with Ergonomic Handle

75 x 25 mm

Designed for general use soft
tissue retraction, the ergonomic
handle allows for a better grip
and less fatigue

75 x 15 mm

6241 [50 x 16 mm]

Overall Length: 8.875" (22,5 cm)
Blade Width: 16 mm
Blade Depth: 50 mm

6242 [75 x 15 mm]

Overall Length: 9" (22,9 cm)
Blade Width: 15 mm
Blade Depth: 75 mm

6243 [75 x 25 mm]

Overall Length: 9" (22,9 cm)
Blade Width: 25 mm
Blade Depth: 75 mm

Non-glare finish featured on
the metal retractor parts.

50 x 16 mm

Rake Retractors
with Ergonomic Handle
Designed for general use
soft tissue retraction

3-Prong

PRODUCT NO’S:

4839 [3-Prong]

Overall Length: 9.5" (24,1 cm)
Rake Width: 13 mm
Rake Depth: 14 mm

4840 [4-Prong]
4-Prong30

WWW.INNOMED.NET   Overall Length: 9.5" (24,1 cm)
Rake Width: 19 mm
Rake Depth: 14 mm   FREE TRIAL ON MOST INSTRUMENTS

Deep Meyerding Retractor
with Ergonomic Handle
A self-retaining soft tissue
retractor for use in hip, knee,and shoulder surgery
PRODUCT NO:

MADE EXCLUSIVELYFOR INNOMED IN

6244

Overall Length: 8.5" (21,6 cm)
Blade Depth: 3.5" (8,9 cm)
Blade Width: 1" (2,54 cm)

Gelpi Retractors

With
Ergonomic
HandleStandard

PRODUCT NO’S:GERMANY

4180 [Standard]

Overall Length: 7.75" (19,7 cm)

4181 [With Ergonomic Handle]
Overall Length: 8.75" (22,2 cm)

Mantis Screwdriver Distractor

Designed by J. Albert Diaz, MD

Designed to help provide stable distraction
across difficult-to-reduce fracturesusing two seated screwdrivers*

Accommodates screwdrivers of varying sizefor use with both small and large fragment systems
Allows for distraction of difficult-to-reduce fractures
without the need to drill additional holes outside of the plate
The plate can be locked with a screw once length has been restored.
PRODUCT NO:

3654

Overall Length: 7.5" (19,1 cm)
Pin Hole Diameters: 4.5, 5.5, & 8.5 mm
Leg & Pin Hole Depth: .7" (17,5 mm)

1.800.548.2362       MAY 2023   *Screwdrivers not included.
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   TRAUMA INSTRUMENTS

31

Joint, Calcaneal andSmall Bone Distractors

Two hole sizes and two arm designs allow
for easier pin size selection and helps with
distraction in a variety of indications
PRODUCT NO’S:OUTSPREAD ARMS

CLOSED ARMS

4210-LB [Large]

4210-LS [Large]

4210-SB [Small]

4210-SS [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

4210-XSD [Extra Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 4.25" (10,8 cm)

LARGE

Outspread andClosed Arms

EXTRA SMALL

Closed Arms only

SMALL

Outspread andClosed Arms

Joint, Calcaneal and Small BoneDistractors with Thumbscrews
Thumbscrew Modification Designed by Kelly McCormick, MD
PRODUCT NO’S:OUTSPREAD ARMS

CLOSED ARMS4215-LB [Large]

4215-LS [Large]

4215-SB [Small]

4215-SS [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)
Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Thumbscrews help prevent theunit from sliding on the pins

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

WITH THUMBSCREWS

Large and Small,
Outspread and Closed Arms

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Joint, Calcaneal, andSmall Bone Compressor

Designed for compression in
fracture and osteotomy procedures
Two hole sizes for ease of pin size selection:.062" (1,6 mm) & .094" (2,4 mm)

SMALL

PRODUCT NO’S:

4210-SC [Small]

Overall Length: 6" (15,2 cm)

EXTRA SMALL

4210-XSC [Extra Small]

Overall Length: 4.25" (10,8 cm)

32

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Joint, Calcaneal, and Small Bone
Compressor/Distractors
with Speed Lock
Speed lock helps allow precise control
and prevents unintended release
Two hole sizes allow for pin size selection.

Large
Closed Arms
with Speed
Lock
Small
Closed Arms
with Speed
Lock

Thumbscrews
help prevent theunit from sliding
on the pins

PRODUCT NO’S:
CLOSED ARMS WITH SPEED LOCK

4216-LS [Large]

OUTSPREAD ARMS WITH

Holes Diameters: For .062" & .094" SPEED LOCK & THUMBSCREWS(1,6 & 2,4 mm) K-wire Pins
4217-LB [Large]
Overall Length: 8" (20,3 cm)
Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
4216-SS [Small]
Overall Length: 8" (20,3 cm)
Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
CLOSED ARMS WITH
Overall Length: 6" (15,2 cm)
SPEED LOCK & THUMBSCREWS

4216-XS [Extra Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 4.5" (11,4 cm)

Extra Small
Closed Arms with
Speed Lock

Large
Outspread Arms
with Speed Lock
& Thumbscrews

Small
Closed Arms with
Speed Lock &Thumbscrews

4217-SS [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Large Pin
Distractor and Compressor

LARGE PINCOMPRESSOR

Larger 1/8" (3,2 mm) pin hole size forextra sturdy distraction or compression
PRODUCT NO’S:

4233 [Large Pin Distractor]

Hole Diameters: For .125" (3,2 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

LARGE PIN
DISTRACTOR

Joint, Calcaneal, and Small
Bone Compressor/Distractor

4234 [Large Pin Compressor]

Hole Diameters: For .125" (3,2 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

1/16" & 3/32"(.062" & .094")
Pin Hole Diameters(1.6 mm & 2.4 mm)

Selection Lever

Selection lever switches the mechanism
from compression to distraction
Simply squeeze the handle one time after
direction selection to engage the mechanism.
Two hole sizes for pin size selection.
PRODUCT NO:

4865-LS

Overall Length: 8.5" (21,6 cm)
Holes For: .062" & .094" (1,6 & 2,4 mm) K-wire Pins

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

33

HFD Compressor/Distractor

PRODUCT NO’S:
SMALL

1834 [Small – All Stainless Steel]

Dial mechanism helps allow precise
control of inserted wires— for maintaining
a position, compressing or distracting

Dimensions: 51 mm x 57 mm
Maximum Arm Opening: 1.35" (3,4 cm)

1834-R [Small w/Radiolucent Arms]
Dimensions: 51 mm x 57 mm
Maximum Arm Opening: 1.35" (3,4 cm)

A .125" (3,2 mm) pin can be used in theholes of the thumbwheel for leverage
Small: Two hole sizes allow for ease of pin size
selection: .045" (1,1 mm) & .062" (1,6 mm)
Large: Two hole sizes allow for ease of pin size
selection: .082" (2,0 mm) & .125" (3,2 mm)
Radiolucent arms are a steam sterilizable
PEEK/Carbon Fiber composite

LARGE

1836 [Large – All Stainless Steel]Overall Length: 4" (10,2 cm)
Maximum Arm Opening: 2.25" (5,7 cm)

1836-R [Large w/Radiolucent Arms]
Overall Length: 4" (10,2 cm)
Maximum Arm Opening: 2.25" (5,7 cm)

Radiolucent arms

Wurapa Small Joint
Compressor and Distractor

Available with two holesizes on each instrument!

Designed by Raymond K. Wurapa, MD

Designed to allow one-handed
manipulation and deployment
once fixation pins are placed
K-wires should be cut short above the pin
guides to allow full access to the operative site

Designed to simplify several
small joint procedures:

PRODUCT NO’S:DOUBLE HOLES:.045" (1,1 mm) & .062" (1,6 mm)

Compressor

1751 [Compressor]

Compresses From: 28 mm
Overall Length: 4.625" (11,7 cm)

1752* [Distractor]

Preparation of small bone non-unions
before bone grafting and fixation
Preparation of small joints for arthrodesis(e.g. partial wrist fusion)
Distract and better evaluate small joints
before determining final management
Useful for intercarpal stabilization while
performing ligament reconstructions(e.g. scapholunate ligament repair/reconstruction)

Distracts to: 46 mm
Overall Length: 4.625" (11,7 cm)

SINGLE HOLE:.045" (1,1 mm) Hole

1753 [Compressor]

Compresses From: 28 mm
Overall Length: 4.5" (11,4 cm)

1754 [Distractor]

Distracts to: 46 mm
Overall Length: 4.5" (11,4 cm)

Distractor

MADE EXCLUSIVELYFOR INNOMED IN*

GERMANY

Faillace Ambidextrous Self-Retaining Retractor
Designed by John J. Faillace, MD

Handle can be rotated away from thesurgeon after insertion if desired
PRODUCT NO’S:

1580 [7 Teeth]

Overall Length: 7.5" (19,1 cm)
Prong Depth: 38 mm
Prong Width: 34 mm

1579 [4 Teeth]

Overall Length: 6" (15,2 cm)
Prong Depth: 38 mm
Prong Width: 18 mm

1579-01 [Small – 4x3 Teeth]
Overall Length: 5.25" (13,3 cm)
Prong Depth: 20 mm
Prong Width: 18 mm / 13 mm

34

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Gurbani Joint Distractor/Compressor
Versatile joint distractor/compressor forarthroscopic or open procedures of foot,byankle, hand, and wrist joints Designed
Naren G. Gurbani, MD
PRODUCT NO’S:

4208-00 [Set]T-wrench helps
provide precise,controlled manipulation

Pin Hole Sizes:.15" (3,5 mm) and.182" (4,5 mm)

Includes: Distractor/Compressor, T-Wrench, and Case

Available individually:4208-01 [Distractor/Compressor Only]

The surgeon puts the pins in the bone, then slides
the holes of the device over the pins and distracts
or compresses—the device can be locked in either
direction. Especially useful for arthroscopy of subtalar,talo-navicular, calcaneo-cuboid, and wrist joints.

Dimensions: 6" w x 5" h (15,2 cm x 12,7 cm)
Distracts up to: 3" (7,6 cm) / Compresses down to: .5" (1,3 cm)

4208-TW [T-Wrench]

Dimensions: 3" w x 3" h (7,6 cm x 7,6 cm)

1025 [Sterilization Case]

HFD Self-Retaining Small Bone Spreader
Versatile spreader featuring narrow
tapered blades which, when together,make a small wedge to enter a tightbone interface or osteotomy
Blades feature a non-aggressive grip
pattern that can be used when spreading
apart bone as well as providing retractionof soft tissue in a smaller wound.
PRODUCT NO:

1829

Overall Length: 4.5" (11,4 cm)
Blade Depth: 28 mm
Blade Width Tapers from: 8 mm to 5 mm

Wurapa Swivel Blade Forearm Retractor
Designed by Raymond Wurapa, MD

Designed for forearm and wrist fracture exposure,the blades swivel for less stress on soft tissue
PRODUCT NO’S:

1646-00 [Set]

Includes Retractor and Two Swivel Blades

Also available individually:

1646-01 [Retractor]

Overall Length: 5.125" (13 cm)
Opens to: 2.5" (6,4 cm)

Swivel-blade technology helps to allow parallel deploymentof retractor blades to maximize wound exposure andminimize edge loading on surrounding soft tissues. Parallel
deployment of the retractor blades also helps prevent
rotation and migration of the retractor during a procedure.

1646-02 [Swivel Blade]

One blade with this product number, two included in set

Width: .9375" (24 mm)
Depth: .75" (19 mm)

Williams Distal Radius Fracture Retractor
Designed to provide excellent exposure
during fracture reduction and plating
PRODUCT NO’S:

1837-L [Left]

For Pins up to .045" (1.1 mm)
Overall Length: 4.5" (11,4 cm)
Blade Depth: 20 mm
Blade Width: 12.5 mm

Designed by
Craig S. Williams, MD
and Eric Dahlinger

Left
Right

1837-R [Right]

For Pins up to .045" (1.1 mm)
Overall Length: 4.5" (11,4 cm)
Blade Depth: 20 mm
Blade Width: 12.5 mm

MADE FOR INNOMED IN

GERMANY

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

35

Dodson Modular Retractor

Set consists of one
ratcheting self-retaining
retractor, two stainlessmini-hohmann
1838-00 [Set] steel
retractor blades, and a
Replacement Parts: stertilization case.

US Patent No. 9,161,745,82

Designed by Mark A. Dodson, MD

PRODUCT NO’S:

Designed to help expose a small to medium sizebone for internal fixation—can be used for distal
radius, ulna, humerus, and fibula fractures

1838-01 [Retractor Only]
Overall Length: 5.5" (14cm)

1838-02 [Blade Only – One]

Allows the limb to be rotated (pronated or supinated) without loss of
exposure. The hohmann retractors have three hole sizes which allow fora variety of positioning angle options using the teeth of the self-retaining
retractor, or can also be positioned in-between the teeth. The hohmann isplaced around the bone, and thus reduces the force on the soft tissues while
increasing exposure. Can be used in the forearm to treat radius and ulna
shaft fractures, humerus fractures, as well as in the leg for fibula fractures.
The optional radiolucent blades are made of a strong, lightweight carbon
fiber PEEK composite material, which is completely radiolucent, helps toprevent from marring component surfaces, and can be steam sterilized.

Overall Length: 5.25" (13,3cm)
Blade Width: 3/8" (9mm)1025 [Sterilization Case Only]* MADE EXCLUSIVELYMADE EXCLUSIVELYFOR INNOMED IN

FOR INNOMED IN

GERMANYSWITZERLAND

PRODUCT NO:
Optional Part — Not Included In Set:

Optional radiolucent
carbon fiber PEEK composite blade

1838-02R* [Radiolucent Blade]
Overall Length: 5.25" (13,3cm)
Blade Width: 3/8" (9mm)

Chung Weitlaner Retractors
Designed by Raymond Chung, MD

3x4 Prongs
Prong lengths of 25 mm and 30 mm
US Patentwith
No. 9,161,745,82
available
either sharp or blunt tips

3x2 Prongs

PRODUCT NO’S:
3x4 Prongs — Blunt Tips

3x4 Prongs — Sharp Tips

PRODUCT NO’S:
2x3 Prongs — Blunt Tips

2x3 Prongs — Sharp Tips

5065-01 [25 mm]

5066-01 [25 mm]

5065 [25 mm]

5066 [25 mm]

5067-01 [30 mm]

5068-01 [30 mm]

5067 [30 mm]

5068 [30 mm]

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)

25 mm

Longer prongs
allow use in asmall, but deep
wound

Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

30 mm

Beard Distal Bicep
Retractor

Designed by David Beard, MD

Designed to help optimize surgical
exposure during anterior single incision
distal biceps tendon reinsertion
The blade design features an anatomically contoured
distal end to hug the radius cortex. The smooth
distal end helps to avoid deep penetration, and thewidth matches the width of the distal biceps tendon
insertion site. The narrow curved handle design helps
to optimize workspace and visualization.

PRODUCT NO’S:

5834-00 [Set – Retractor & Two Blades]
Available Individually:

5834-01 [Blade] 1 blade with this product number
Overall Length: 6.375" (16,2 cm)
Width: .625" (16 mm)

5834-02 [Self-retaining Retractor]
Overall Length: 7.5" (19,1 cm)

Sold as a set, or available individually for replacement.

Beard Distal Radius
Wide Hohmann Retractor
Designed by David Beard, MD

Designed for distal radius anddiaphyseal fracture exposure, thewide blade design helps to protect
soft tissues, and the curved
handle helps provide improved
access and visualization

PRODUCT NO:

5837-01

Overall Length: 5.375 (13,7 cm)
Blade Width: 1" (25 mm)

36

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Lawton Distal Radius Mini Frame & Blade Set
Designed by Jeffrey Lawton, MD

Designed for self-retaining exposure for distal radiusand other small bone fractures

Short Blade Depth

Small & Large
Blade Depth

Set includes: (1) Frame,(2) Short Blades, (2) Small Blades.
Optional Large Blade available separately.
PRODUCT NO:

1578-00 [Set]
Set Includes / Available Individually:

1578-01 [Mini Frame]

Dimensions: 3" x 2.5" (7,6 x 6,4 cm)

1578-02 [Mini Short Blade](2) included in set, (1) with this product number
Overall Length: 2.5" (6,4 cm)
Blade Width: .625" (16 mm)
Blade Depth: .875" (22 mm)

1578-03 [Mini Small Blade](2) included in set, (1) with this product number
Overall Length: 2.625" (6,7 cm)
Blade Width: .625" (16 mm)
Blade Depth: 1.125" (29 mm)
Optional Blade / Not Included In Set:

1578-04 [Mini Large Blade]
Overall Length: 2.5" (6,4 cm)
Blade Width: .935" (24 mm)
Blade Depth: 1.125" (29 mm)

Short Blade

Small Blade

Optional
Large Blade

OrthoLucent™ Mini Hohmann Retractors
Designed by Jeffrey Lawton, MD

Radiolucent,lightweight retractors
The carbon fiber PEEK material is strong,lightweight, completely radiolucent, canbe steam sterilized, and helps to preventfrom marring component surfaces.
PRODUCT NO’S:

1594-R [8 mm Blade]

Overall Len gth: 6.875" (17,5 cm)
Blade Width: 8 mm

PRODUCT NO’S:

1159 [Standard Sharp Rake]
Overall Length: 4.5" (11,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1161 [Standard Blunt Rake]
Overall Length: 4.5" (11,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1162 [Standard Senn]

Overall Length: 4.5" (11,4 cm)
Blade Width: 6 mm
Blade Depth: 16 mm

1597-R [16 mm Blade]

Overall Length: 6.875" (17,5 cm)
Blade Width: 16 mm

Chung T-Handle Retractors

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

Designed by Raymond Chung, MD

Designed with a T-handle for easier holding andto help reduce finger and thumb fatigue

Rake

Sharp Rake

Standard
Shaft Length

Senn

Blunt Rake

1159-01 [Extended Sharp Rake]

Senn

Overall Length: 5.625" (14,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1161-01 [Extended Blunt Rake]
Overall Length: 5.625" (14,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1162-01 [Extended Senn]
Overall Length: 5.625" (14,4 cm)
Blade Width: 6 mm
Blade Depth: 16 mm

Extended
Shaft Length

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

37

Star Bit Driver Set

T10

Helps eliminate the opening ofmultiple sterile packs when aspecific size of star bit is needed
PRODUCT NO’S:

T25

5194-00 [4 Star Bits w/Handle & Case]
5194-01 [4 Star Bits w/Case only]

T20

Also sold individually:

S0113 [Universal 4" (10,2 cm) Handle]
5194-10 [T10 with A/O End]5194-15 [T15 with A/O End]5194-20 [T20 with A/O End]5194-25 [T25 with A/O End]
9003 [Case]

T15

T15

T20

T25

Helpful during revision total joint surgery. Set
consists of four star bits — T10, T15, T20, & T25,
a handle which accommodates any of the above
bits, and a sterilization case. The drive end (A/O) is
designed for easy and quick engagement with theuniversal instrument handle. The ergonomic, modular
handle has two connection points, allowing for bothstraight and T-handle orientations.

T10
Set in Storage Case

Universal Screw Removal Instrument System

Designed to help remove a variety of screws—solid and cannulated:
stripped hex screws, buried screws, partial screws with broken screw heads

The drive end (A/O) is designed for easy and quick
engagement with the universal instrument handle.
Screw Extractors

Unique thread design accommodates removal of stripped screws.
The instrument “locks” into the screw head and allows removal once
engaged. Designed to be used in a counter-clockwise direction.

Designed to fit over submerged screws for extraction with minimalbone loss. Extraction is enhanced by the unique tooth design.
Designed to be used in a counter-clockwise direction.

Hex Drivers

Cannulated Hex Drivers

Solid shaft in all standard hex sizes.

Universal Extractor

Designed to remove screws with heads partially or completely
missing. The cone shaped head fully engages the remaining
screw and optimizes the force needed for removal. The bolt
is disposable and locks into place using a unique thread
design. Designed to be used in a counter-clockwise direction.

PRODUCT NO’S:

Trephines

Four sizes with a cannulated shaft for easier removal of buried screws.

Screwdrivers

Standard cruciform screwdrivers in large,
small, and mini, and single slot.

Cannulated Drive
Extension

Used when a longer
instrument shaft is desired.

Push button Quick-connect
release mechanism

S0010-00 [Complete System with Case]
Individual/Replacement Parts

S0113 [Universal 4" (10,2 cm) Handle]
S0128 [1.5 mm Screw Extractor]
S0116 [2.5 mm Screw Extractor]
S0130 [3.5 mm Screw Extractor]
S0117 [1.5 mm Hex Driver]S0114 [2.5 mm Hex Driver]S0115 [3.5 mm Hex Driver]S0132 [4.0 mm Hex Driver]S0133 [5.0 mm Hex Driver]S0136 [2.5 mm Cannulated Hex Driver]S0137 [3.5 mm Cannulated Hex Driver]S0138 [4.0 mm Cannulated Hex Driver]S0139 [5.0 mm Cannulated Hex Driver]S0118 [Large Cruciform Screwdriver]S0119 [Small Cruciform Screwdriver]S0141 [Mini Cruciform Screwdriver]S0120 [Single Slot Screwdriver]
S0121 [2.2 mm Trephine]
S0122 [3.2 mm Trephine]
S0123 [4.2 mm Trephine]
S0124 [4.7 mm Trephine]
S0125 [7.2 mm Trephine]
S0127 [Universal Extractor – Shaft Only]
S0127-01 [Large Extraction Bolt Body]
S0127-03 [Small Extraction Bolt Body]
S0127-04 [Extractor Wrench]S0129 [Pick]
S0140 [Cannulated Drive Extension]
9017 [Screw Removal Case Only]
Case Dimensions: 21" x 9.5" x 2.25
	   (53,4 x 24,1 x 5,7 cm)

Ergonomic, modular handle with twoconnection points allows for bothstraight and T-handle orientations

38

Extractor
WrenchUniversal Instrument Handle

The single handle allows the surgeon to decide
which direction is most efficient and comfortable.
The quick-connect release mechanism allows forquick interoperative exchange.

Pick

Used to remove
fragments andbone or tissue
from screw head.

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Cheng Screw Removal and Bone Trephine Set

5 mm

6.5 mm

8 mm

9 mm

10 mm

11 mm

Trephine Sizes

Designed by Edward Cheng, MD

K-wire not included.

Six trephine sizes with reverse thread teeth
designed to help with removal of screws
with minimal bone loss, as well asgathering of core bone samples forbiopsy or core decompression
Can be used with the T-handle
or with power.

For Screw Removal

The trephine ends are designed to fit over embedded screws for extraction
with minimal bone loss. Six sizes available — internal diameters of 5 mm, 6.5 mm,
8 mm, 9 mm, 10 mm, and 11 mm. The T-handle allows for precise, controlled use.

1426-00 [Complete Set with Case]
Set Includes/Available Separately:

1426-01 [5 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-02 [6.5 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-03 [8 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-05 [9 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-06 [10 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1426-07 [11 mm Internal Diameter]
Overall Length: 7.125" (18,1 cm)

1425-14 [Handle Assembly]

Dimensions: 4" x 2" (10,2 cm x 5,1 cm)

For Core Bone Sampling

Cannulated T-handle and trephines allow use of a standard 1.6 mm (.062") threaded K-wire to help facilitate
grasping and removal of a core bone sample for biopsy or core decompression. Variety of core diameters yields
bone samples of sufficient size for pathology. K-wire not included.

small & large single slot
cross & cruciate
3.5 mm & 4.5 mm hex
small & large phillips#6 & #8 star
#10 & #15 star
#20 & #25 star

1025 [Sterilization Case]
Replacement Part:

1425-14-B-COMP [Handle Retaining Screw]

Universal Screwdriver Set
Helps eliminate the openingof multiple sterile packs whena specific size or style ofscrewdriver is needed

Helpful during revision total joint surgery wherescrews have been used, removal of bone plates,
fracture fixation screws or bone graft screws.
PRODUCT NO’S:

Set consists of one handle
and one sterilization/storage
case, plus seven double
ended screwdriver bits:

5195 [Complete Set with Case]
Also sold individually

5195-01 [Handle]
5195-02 [Straight (single slot)]
Large: 7 x 1.5 mm, Small: 5 x 1 mm

small & large single slot
cross & cruciate
3.5 mm & 4.5 mm hex
small & large phillips#6 & #8 star
#10 & #15 star
#20 & #25 star

5195-03 [Cross/Cruciate]
Large: 7 mm, Small: 6 mm

5195-04 [Hex]Large: 4.5 mm, Small: 3.5 mm

5195-05 [Phillips]

Large: 4 mm, Small: 3.5 mm

5195-08 [Small Star: #6 & #8]
5195-06 [Medium Star: #10 & #15]
5195-07 [Large Star: #20 & #25]

Torx/Hex Adapter Set

PRODUCT NO’S:

8003-00 [Set – One Each]
Set Includes/Available Separately:

Designed by Stephen M. Walsh, MD

8003-01 [Torx Bit to Hex Driver Adapter]

Designed for conversion
of a 3.5 mm screwdriver
Especially helpful when an
articulated, universal joint driver isneeded (i.e. acetabular screws)

PRODUCT NO’S:

Overall Length: .6" (1,54 cm)

8003-02 [Hex Bit to Torx Driver Adapter]
Overall Length: .6" (1,54 cm)

Hex Bit to Torx Driver Adapter

1.800.548.2362       MAY 2023   

Torx Bit to Hex Driver Adapter   TRAUMA INSTRUMENTS

39

Screw/Pin Removal Locking Pliers
Unique jaw designed to solidly grip
and clamp onto a screw head, broken
screw, or pin for removal
NEW Small
Jaw End & Bite

Standard
Jaw End & Bite

Designed to securely grab pins
as small as 1.4 mm (.055")
up to 2.4 mm (.095")

Designed to securely grab
larger pins, screw heads, or
broken screws

Standard

PRODUCT NO’S:

S0142 [Standard]

Overall Length: 7.875" (20 cm)
Jaw Width at End: 4 mm

Reduced jaw size for smallerscrews, pins and incisions

S0142-01 [Small]

Overall Length: 7.875" (20 cm)
Jaw Width at End: 4 mm

Small

Screw Extractor with Speed Lock
Designed by Khaled Sarraf, MD & Konstantinos Doudoulakis, MD

Universal extractor designed to accommodate a large
range of screws and screw heads from 3.95 to 9.5 mm
Can also be used to help with removal of other devices
that may require a twisting universal locking gripper.
PRODUCT NO:

2021

Overall Length: 9.25" (23,5 cm)
Jaw Width: 11 mm
Jaw Length: 5 cm

Screw Removal Pliers
Jaw designed to grasp
onto a screw or screw
head to help in removal
PRODUCT NO:

2020

Overall Length: 8 (20,3 cm)

Long Jaw Needle Nose Pliers
PRODUCT NO:

1833

Overall Length: 7" (17,8 cm)
Jaw Length: 2.25" (5,7 cm)
Jaw Width Tapered from: 8 mm to 1.5 mm
Jaw Height Tapered from: 12 mm to 2.5 mm
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY40

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTSSTANDARD LARGEPRODUCT NO’S:

OrthoVise™ Length: 10" (25,4 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3980-01 with Attachment Bolts (two sides & end)
without Slap Hammer
3981
without Attachment Bolts
without Slap Hammer
with End Attachment Nut that accepts aStandard Slap Hammer (#3925 or 3926)

3980

#3980

Side Attachment Bolts

Allows a Large OrthoVise™
Slap Hammer (#3950) to be
attached to the side of the device.

#3980-01

Available on Standard Large and Long Nose
Large OrthoVise™ with Attachment Bolts only.
(Cannot be added on later.)#3981

LONG NOSE LARGEPRODUCT NO’S:

OrthoVise™ Length: 12" (30,5 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3965-01 with Attachment Bolts (two sides & end)
without Slap Hammer

3965

#3965

OrthoVise™

#3965-01

U.S. Patent #D398,208

LONG NOSE LARGE BENT JAW
PRODUCT NO’S:

OrthoVise™ Length: 11.5" (29,2 cm)
with Attachment Nut (end)with Standard Slap Hammer (#3925)
3966-01 without Slap Hammer
with Attachment Nut (end) that accepts aStandard Slap Hammer (#3925 or 3926)

3966

#3966
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY#3966-01

Made of stainless steel
Models equipped with attachment bolts
allow a slap hammer to be attached to theend, as well as to either side of the large
OrthoVise™ (except Bent Jaw models), forgreater adaptability
Bent Jaw models are not available withside attachment bolts, but have an end
attachment nut to accept a Standard
Slap Hammer (#3925 or #3926)
A different size slap hammer is used forthe large and small sizes of OrthoVise™
Slap Hammers are designed with ahammer plate for the additionaluse of a mallet if desired

STANDARD SMALLPRODUCT NO’S:

OrthoVise™ Length: 8" (20,3 cm)
without Attachment Bolt
without Slap Hammer
3985-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3985-T with Attachment Bolt (end)without Slap Hammer

#3985

3985

#3985-01#3985-T

LONG NOSE SMALLPRODUCT NO’S:

OrthoVise™ Length: 9.5" (24,1 cm)
without Attachment Bolt
without Slap Hammer
3975-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3975-T with Attachment Bolt (end)without Slap Hammer

3975

SLAP HAMMERS#3975#3975-01#3975-T

PRODUCT NO’S:

3950 [Slap Hammer for Large OrthoVise]For use with 3965's, 3980's, 3981
Overall Length: 16.5" (41,9 cm)

For Large
OrthoVise#3950

For use with: 3975's, 3985's
Overall Length: 8.75" (22,2 cm)

For Small
OrthoVise#3955

For use with: 3966's
Overall Length: 16" (40,7 cm)

Standard
with 16" Rod#3925

For use with: 3966's
Overall Length: 16" (40,7 cm)

Easy Grip Standard
with 16" Rod3955 [Slap Hammer for Small OrthoVise]3925 [Standard Slap Hammer w/16" Rod]3926 [Easy Grip Slap Hammer w/16" Rod]#3926

THREADED ADAPTERS
PRODUCT NO’S:

3980-02 [Small Adapter]

Changes Male End of a Slap Hammer to Female

Small Adapter

For use with 3965's, 3966's, 3980's, 3981

Threaded Large

For use with: 3975's, 3985's

Threaded Small

3980-03 [Threaded Adapting Screw – Large]

3985-03 [Threaded Adapting Screw – Small]

1.800.548.2362       MAY 2023   

Female/Female
Adapter Converts
from Male/Male

Small Adapter allows a Standard Slap Hammer (#3925 or #3926)
to be used with any Large OrthoVise™ with Attachment Bolts

Threaded Adapting Screws can be used to append the corresponding size
OrthoVise™ with an Attachment Bolt for use with a Slap Hammer   TRAUMA INSTRUMENTS

41

PRODUCT NO’S:

7653-00 [Set w/Case]

Lawton Screw Extractors

Individual Parts:

7653-01 [1.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

Designed by Jeffrey Lawton, MD

Designed to help extract mini and micro
fragment screws; small cannulated
screws; or headless screws

7653-02 [2.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

7653-03 [3.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

1025 [Sterilization Case]

Lawton Broken Screw Extractor
Designed by Jeffrey Lawton, MD

Designed to help remove broken or
stripped screws (1 mm-2 mm)
PRODUCT NO:

7653-04

Overall Length: 4" (10,2 cm)
Handle Width: 3" (7,6 cm)

Mengato Depth Gauge
Designed by Richard Mengato, MD

Ring-handed design with 3 rings gives 3-point
grip for ease of holding and manipulation

PRODUCT NO:

Allows for superior gauge control and manipulation, to advance, engage
and maintain the hook on the distal cortex by levering the probe
against the bone hole and keeping gentle tension on the hook.

1139

Overall Length – Contracted: 7.125" (18,1 cm)
Overall Length – Extended: 9.125" (23,2 cm)
Gauge: 0 to 50 mm

Depth Gauge

Designed to help provide measurement of thedepth/length of any bone hole for properscrew length determination
PRODUCT NO:

8015

Overall Length: 7.625" (19,4 cm)
Scale: From 0 to 48 mm

42

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

StainlessImpactor Sizes

DelrinImpactor Sizes

Modular Impactor Set

PRODUCT NO:

5370 [Complete Set]

Makes multiple impactor heads
easily visible and available
Designed to have available to the operating
surgeon multiple types of impactors utilizing one
handle. The rack uses less space and allows thesurgeon to quickly see the designs available. The
impactors are supplied with stainless steel tips forbone and delrin tips which can be used against
an implant for slight placement adjustments.

Included In Set / Also Available Individually:

5370-01 [Rectangular Tip 11 mm x 4 mm Steel]
5370-02 [Oval Tip 13 mm x 8 mm Steel]
5370-03 [Crescent Tip 12 mm x 5 mm Steel]
5370-04 [Square Tip 9 mm x 9 mm Steel]
5370-05 [Round Tip 15 mm Steel]
5370-06 [Round Tip 12 mm Steel]
5370-07 [Round Tip 9 mm Steel]
5370-19 [Set Base] Base Diameter: 3.5" (8,9 cm)
5370-D1 [Rectangular Tip 11 mm 4 mm Delrin]
5370-D2 [Oval Tip 13 mm x 8 mm Delrin]
5370-D3 [Crescent Tip 12 mm x 5 mm Delrin]
5370-H [Modular Handle]
Overall Length: 8" (20,3 cm)
Grip Length: 4.5" (11,4 cm)

PRODUCT NO’S:

Overall Length: 9" (22,9 cm)
Shaft Diameter: 9 mm

5331 [11 x 4 mm Rectangle]
5332 [12 x 7 mm Rectangle]
5333 [12 mm Tapered]5334 [9 mm Square]5335 [15 mm Round]5336 [12 mm Round]5337 [9 mm Round]

Ortho
Impactors
Universal Bone Grafting/Impacting Forceps
Designed by J. A. Amis, MD

Bone graft can be grasped, placed & impacted
without changing hands or instruments
The forceps are designed with grasping ends for delivery of bone graft. When thegraft is in place, the forceps are closed, which forms the ends into an impacting
punch. A striking platform is attached to the end of the forceps for tapping andtamping the graft. Four end diameters are available in two lengths.
PRODUCT NO’S:
Short: 6" (15,2 cm) Length

1/8"
3/16"
1/4"
5/16"
(3,2 mm) (4,8 mm) (6,3 mm)(8 mm)
Diameter ends at actual size (closed forceps)

When the forceps areclosed, they form into
an impacting punch

5010-01 1/8" (3,2 mm) Diameter End
5010-02 3/16" (4,8 mm) Diameter End
5010-03 1/4" (6,3 mm) Diameter End
5010-04 5/16" (8 mm) Diameter End
Long: 10" (25,4 cm) Length

5050-01 1/8" (3,2 mm) Diameter End
5050-02 3/16" (4,8 mm) Diameter End
5050-03 1/4" (6,3 mm) Diameter End
5050-04 5/16" (8 mm) Diameter End
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

43

Rogozinski Locking Needle Driver/Scissors

Designed with a quick lock & release handle, can drivea needle and cut a suture without changing instruments
Scissors
Holding
Tips

PRODUCT NO’S:

3083 [Standard]

Designed by
Chaim Rogozinski, MD

3084 [Large]

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 6.5" (16,5 cm)

Overall Length: 7.75" (19,7 cm)

GERMANY

Orthopedic Needle
Holder/Scissors

Scissors
Holding
Tips

Drive a needle and cut a suture
without changing instruments
PRODUCT NO’S:
Standard Tips

MADE FOR INNOMED IN

GERMANY

3070 7.0" (17,8 cm)
Tungsten Carbide Tips

3055 5.5" (14 cm)
3065 6.5" (16,5 cm)
3075 7.0" (17,8 cm)

Longer sizes arehelpful in orthopedics

Large T-Handle Fixed Drivers

Large easy grip soft silicone handleddrivers help provide a sturdy non-slip grip

Zimmer Hall
Quick-connect

PRODUCT NO’S:

8248 [Zimmer Hall Quick-connect]

HudsonQuick-connect

Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

8248-01 [Zimmer Hall Reverse Quick-connect]
Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

8249 [Hudson Quick-connect]

Overall Length: 6.75" (17,1 cm)
Overall Length with Pin In Handle: 11.5" (29,2 cm)

Zimmer Hall
Reverse Quick-connect

Zimmer Hall
Quick-connect

HudsonQuick-connect

The two standard Quick-connect
models release by pulling the collar
backward, while the Reverse Quickconnect model is designed to have thecollar be pushed forward for release.

T-Handle Chuck & Key
PRODUCT NO’S:

8247-00 [T-Handle Chuck & Key]8247-01 [T-Handle Chuck Only]
8247-02 [Chuck Key Only]

44

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Clear Vision Debris Shield
Designed by R. Barry Sorrells, MD

Provides a degree of restriction fromflying debris or liquid during surgery
Held between the surgical site and the operating
personnel, the shield provides a clear undistorted
view, while helping to protect the patient andpersonnel from possible contamination. The shield is
autoclavable and gas sterilizable in a flat position.
PRODUCT NO:

8031-01

Dimensions: 8" x 10.25" (20,3 cm x 26 cm)(Dimensions do not include the handle)

Adson Forceps
with Cobb Elevator End
Designed by Oscar Castro-Aragon, MD

Has the advantages of having a
Cobb tip at the end of an Adson forceps
Allows the opportunity to do soft tissue dissection,cleaning of the bone or bone fragments in a fracture,
push bone fragments to hold a reduction in a fracture,
separate soft tissue, and turn it around to pick up tissue
without having to switch instruments back and forth.

PRODUCT NO:

1166

Overall Length: 4.75" (12,1 cm)
Tip Width: 2.4 mm (2,4 mm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Rudisill Locking
Small Bone Reduction Forcep
Designed by Ed Rudisill, MD

For reduction of hand phalanx
and metacarpal fractures
PRODUCT NO:

MADE EXCLUSIVELYFOR INNOMED IN

2017

Overall Length: 4.875" (12,4 cm)

GERMANY

Faillace Bone Impact/Graft Forceps
Design modification by John J. Faillace, MD, FAAOSLong vertical grooves at the tip aredesigned to deliver graft into a small
space, where a freer elevator can beused to push the graft down into thespace, then the closed flat end canbe used to tamp down the graft

1.800.548.2362       MAY 2023   

PRODUCT NO:
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   TRAUMA INSTRUMENTS

5011

Overall Length: 5" (12,7 cm)
Tip Diameter When Closed: 3,2 mm

45

Gelbke Cobb Elevator with Suction

Designed by Martin K. Gelbke, MD

Designed to be used during exposure of the posterior
spine, as well as for pelvic and acetabular trauma cases
PRODUCT NO:

3433

Overall Length: 12.75" (32,4 cm)
Cobb End Width: 18 mm (.7")
Shaft plus Head Length: 5.5" (15 cm)

Cobb Elevators

Two Sizes Available With or Without Teeth
Ultra hard titanium nitride coating helps
to extend blade life by increasing surface
hardness, prolonging sharpness, andresisting chemicals and corrosion.

Available with orwithout teeth

PRODUCT NO’S:
WITH TEETHWITHOUT TEETH

3432 [1/2" with Teeth]

3436 [1/2” without Teeth]

3434 [1" with Teeth]

3438 [1” without Teeth]

Overall Length: 11" (27,9 cm)
Blade Width: 1/2" (13 mm)
Overall Length: 11" (27,9 cm)
Blade Width: 1" (25,4 mm)

Overall Length: 11” (27,9 cm)
Blade Width: 1/2” (13 mm)
Overall Length: 11” (27,9 cm)
Blade Width: 1” (25,4 mm)

Bradley
Periosteal Elevator
Designed by Gary W. Bradley, MD
PRODUCT NO’S:

4719 [1/2"]

Overall Length: 11" (27,9 cm)
Blade Width: .5" (13 mm)

4720 [3/4"]

Overall Length: 11" (27,9 cm)
Blade Width: .75" (19 mm)

PRODUCT NO’S:

3450 [Curved]

Overall Length: 7.5" (19,1 cm)
Handle Length: 4.5" (11,4 cm)
Blade Size: 16 x 13 mm

Periosteal Elevator

3455 [Straight]

Designed for better control

Overall Length: 7.75" (19,7 cm)
Handle Length: 4.5" (11,4 cm)
Blade Size: 19 x 14 mm

46

Designed with a curved end for easieruse, and sharper sides for ease ofelevating and stripping. The handle isdesigned for better control.

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Extended Scalpel Handle

Designed by Richard Pelliccio, MD

Long thin scalpel handle used with knife blade
to make a skin incision and cut throughfascia to help seat trocars to bone
#10 blade normally used but choice of blade isat surgeons’ discretion. Blade not included.
PRODUCT NO:

3022

Overall Length: 18.9" (48 cm)
Handle Length: 5.5" (14 cm)
Shaft Diameter: .25" (6,35 cm)

Laser mark on shaft allows visual orientationof the blade when passing through a cannula.

Ortho Suction Tube
Designed by T. Eickmann, MD

Very effective for suction and minor retracting
Helps eliminate plugging due to bone, cement fragments, blood clots, etc.
PRODUCT NO:

5465

Overall Length: 9.25" (23,5 cm)
End Hole Dia.: 1 mm
Side Hole Dia.: 1.5 mm

Beicker Curette Suction Device
Designed by Clint Beicker, MD

Designed to help visualization of afracture site within a fracture hematoma
Also useful for arthroscopic curettage of osteochondral lesions.
PRODUCT NO:

4231

Overall Length: 10.5" (26,7 cm)
Curette Cup: 7.5 mm x 5.5 mm

Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

1.800.548.2362       MAY 2023      TRAUMA INSTRUMENTS

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:

47

PRSRT. STD.
U.S. POSTAGE
PAID
GRAND RAPIDS, MI

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 748

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

Mantis Screwdriver Distractor

Designed by J. Albert Diaz, MD

Designed to help provide stable distraction
across difficult-to-reduce fracturesusing two seated screwdrivers*

Accommodates screwdrivers of varying sizefor use with both small and large fragment systems
Allows for distraction of difficult-to-reduce fractures
without the need to drill additional holes outside of the plate
The plate can be locked with a screw once length has been restored.

PRODUCT NO:

3654

Overall Length: 7.5" (19,1 cm)
Pin Hole Diameters: 4.5, 5.5, & 8.5 mm
Leg & Pin Hole Depth: .7" (17,5 mm)*Screwdrivers not included.
MADE EXCLUSIVELYFOR INNOMED IN

GERMANYINNOMED, INC

Innomed-Europe LLC

TOLL FREE 1.800.548.2362

Tel 912.236.0000
Fax 912.236.7766

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74

Innomed-Europe GmbHISO 13485:2016

103 Estus Drive
Savannah, GA 31404

Villingen-Schwenningen,Deutschland
Tel 0049 (0) 7720 46110 60

innomed.net
info@innomed.net© 2023 Innomed, Inc.
All Rights Reserved

05
23

www.innomed-europe.com
orders@innomed-europe.com

INNOMED.NET

info@innomed.net