JANUARY

2024

OR T HOPEDIC INS T RUMEN T S
Featuring®

instruments

Tibiofemoral Offset Caliper
Page 12

Proximal Tibia Resection
Shark Clamp
Page 6

Malleable
Bone Tamps
Page 18

Wilson Condylar Gauge
Page 13

Offset Osteotomes
Page 23

Knee Instruments
1.800.548.2362   INNOMED.NET

Andrews Modified
Tibial Fragment Grasper
Designed by Scott Andrews, MD

Designed to help remove tibial bone during
unicondylar and total knee arthroplasty
PRODUCT NO:

1721

Overall Length: 10" (25,4 cm)
Jaw Dimensions: 1.44" x .72" (36,6 cc x 18,3 mm)
Lower Jaw Thickness: 1 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Rosenstein Tibial
Fragment Grasper for UKA
Designed by Alexander D. Rosenstein, MD

Designed to help remove the tibial
bone fragment in one piece during
Unicompartmental Knee Arthroplasty

The narrow grasper with its thin lower jaw is inserted under
the femoral condyle, helping to secure the tibial fragment
throughout it’s entire length, and to remove the fragment
without breaking it. The angled design helps keep thesurgeon’s hands out of the way and facilitates visualization.
PRODUCT NO:

1720

Overall Length: 10” (25,4 cm)
Jaw Dimensions: 1.44” x .72” (36,6 mm x 18,3 mm)
Lower Jaw Thickness: .05” (1,2 mm)

PRODUCT NO:

3685

Overall Length: 10" (25,4 cm)
Calibrations: 0 to 26 mm

MADE EXCLUSIVELYFOR INNOMED IN

Universal Calibrated
Tibia/Patella Clamp

GERMANY

Designed by S. David Stulberg, MD

Designed to be used to remove a tibia wedge,
helps in everting the patella, and calibrations help in
measuring the thickness of the patella and tibia wedges

Fracchia Tibia/Patella
Clamp with Speed Lock
Designed by Michael J. Fracchia, MD & S. David Stulberg, MD

Designed to be used toremove a tibia wedge, andhelps in everting the patella
Speed lock helps allow precise control
and prevent unintended release.

2

PRODUCT NO:

3645

Overall Length: 10" (25,4 cm)

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Sidhu Tibia Clamp
Designed by Kuldeep Sidhu, MD

Designed to be used to securely grasp
and remove an entire tibial wedge
The tapered lower pad slides under the cut tibial wedge
without first having to use wedges, then, clamping allows
the spikes in the upper pad to securely grasp the entire
tibial wedge for easy removal.
PRODUCT NO:

3643

Overall Length: 10.25" (26 cm)
Pads: 60 mm x 30 mm
Spike Length: 7,5 mm

Andrews Modified Tibial Wedge Clamp
Designed by Scott Andrews, MD and Kuldeep Sidhu, MD

Designed to help remove the cut tibial
bone during total knee procedures
The bone is held by the spikes which helps it to comeout in one piece, and also helps with release of soft
tissues from the bone.
PRODUCT NO:

3642

Overall Length: 10.25" (26 cm)
Pads: 60 mm x 30 mm
Front Spike Length: 14 mm
Back Spike Length: 7,5 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Rosenstein Forked UKA Tibial Fragment Grasper
Designed by Alexander D. Rosenstein, MD

Used to help remove the tibial bone fragment during UKA,
the forked upper jaw design helps the instrument to
fit around a femoral condyle while the thin lower
jaw slips through the osteotomy site

Available in two sizes:
Large designed to fit large knee joints, andSmall to fit small and medium knee joints.
The reverse-angled teeth under the upper jaw firmly grip the tibial fragment through its entire length,
allowing removal of the fragile wafer of tibial bone without breaking it. This unique design helps deploy
the instrument in tight medial or lateral compartments of the knee joint. The angled design keeps thesurgeon's hands out of the way and facilitates visualization.
PRODUCT NO’S:

1720-02 [Large – 23 mm Jaw] 1720-03 [Small – 18.5 mm Jaw]
Overall Length: 10” (25,4 cm)
Jaw Width: 23 mm
Upper Jaw Inside Width: 15.4 mm

1.800.548.2362       JANUARY 2024   

Overall Length: 9.33" (23,7 cm)
Jaw Width: 18.5 mm
Upper Jaw Inside Width: 10.8 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   KNEE INSTRUMENTS

3

Intraarticular Tissue Grasper/Rongeur
Used to securely grasp tissue or
can be used to rongeur tissue
Small Grip Handle
PRODUCT NO’S:

1790-01-SG [5" Small Grip]

9" Small Grip #1790-03-SG

Overall Length: 8" (20,3 cm)
Shaft Length: 5" (12,7 cm)

1790-03-SG [7" Small Grip]

7" Small Grip #1790-03-SG

Overall Length: 10" (25,4 cm)
Shaxft Length: 7" (17,8 cm)

1790-02-SG [9" Small Grip]
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

5" Small Grip #1790-01-SG

Designed with the grip closer
together for easier gripping andto help reduce hand fatigue

Standard Handle

9" Standard Grip with Locking Ratchet #1791-02

PRODUCT NO’S:

1790-01 [5" Standard]9" Standard Grip #1790-02

Overall Length: 8" (20,3 cm)
Shaft Length: 5" (12,7 cm)

1790-03 [7" Standard]7" Standard Grip #1790-03

Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

1790-02 [9" Standard]

Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

5" Standard Grip #1790-01

1791-02 [9" Locking Ratchet]Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

Sure Grip Soft Tissue Grasper

9" Sure Grip #3645-03

Designed by Andrew Glassman, MD

Enables the surgeon to securely grasp
soft tissue structures within the knee
Incorporates a 3 mm spike into its upper jaw with amatching recess in the lower jaw, enabling the surgeon
to securely grasp soft tissue structures within the knee.
Particularly useful for grasping the posterior horn ofeither the medial or lateral meniscus. Also useful when
excising the cruciate ligaments, capturing loose bodies,holding the retinaculum during patellar preparation,and grasping the capsule during wound culture.

7" Sure Grip with Locking Ratchet #3646-02

7" Sure Grip #3645-02

5" Sure Grip #3645-01

PRODUCT NO’S:

3645-01 [5"]3646-02* [7" with Ratchet]3645-02 [7"]3645-03 [9"]Overall Length: 8" (20,3 cm)
Shaft Length: 5" (12,7 cm)
Spike Depth: 3 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)
Spike Depth: 3 mm

4Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)
Spike Depth: 3 mm

Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)
Spike Depth: 3 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY*INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Shark Tooth Graspers
Designed by Luis Ulloa

Helpful in removing the labrum, and osteophytes around the acetabulum andaround the glenoid. Also helps to remove meniscus, osteophytes and loose bodies.
Helps facilitate working through a small incision without disrupting vision.

Sharp teeth help grasp
onto tissue and bone

9" Small Grip #1799-SG

Small Grip Handle
7" Small Grip #1798-SG

PRODUCT NO’S:

1798-SG [7" Small Grip]

Designed with the grip closer
together for easier gripping andto help reduce hand fatigue

Jaw Size: 6 mm x 10 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

1799-SG [9" Small Grip]
Jaw Size: 6 mm x 10 mm
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

12" Standard Grip #1796

Standard Handle
9" Standard Grip #1799

PRODUCT NO’S:

1797 [5" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 8" (20,3 cm)
Shaft Length: 5" (12,7 cm)

7" Standard Grip #1798

1798* [7" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

1799* [9" Standard]5" Standard Grip #1797

Jaw Size: 6 mm x 10 mm
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

1796 [12" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 15" (38,1 cm)
Shaft Length: 12" (30,5 cm)* MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Cartilage Graspers

8" Shark Teeth #1779

Designed by Luis Ulloa

Shark Tooth Jaw

Helps to grasp
and hold cartilage,
tendons, soft tissues
and loose bodies

5" Shark Teeth #1777

Shaft allows for use in
narrow spaces.
6" Saw Teeth #1785
PRODUCT NO’S:PRODUCT NO:

1785 [Saw Teeth]

Saw Tooth Jaw

Shaft Length: 6" (15,2 cm)
Overall Length: 9.25" (23,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       JANUARY 2024   

1777 [5" with Shark Teeth]
Shark tooth design modification by Michael Soudry, MD

GERMANY   KNEE INSTRUMENTS

Shaft Length: 5" (12,7 cm)
Overall Length: 8.25" (21 cm)
Jaw Bite: 2 mm x 6.5 mm

1779 [8" with Shark Teeth]
Shaft Length: 8" (20,3 cm)
Overall Length: 11.25" (28,6 cm)

5Proximal Tibia Resection
Shark Clamp
Designed by Shara Diers, PA-C

Designed to help grasp and remove
the cut proximal portion of the tibia
during total or uni knee arthroplasty

PRODUCT NO:

3651

Overall Length: 7" (17,7 cm)
Platform Width: .79" (2 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Tissue Graspers with Shark Teeth

Up Angled Jaw #1784-01

Designed by Luis Ulloa

Shark teeth help to grasp
on to tissue and bone
Shaft allows for use in narrow spaces.

Straight Jaw #1784-02

PRODUCT NO’S:

1784-01 [Up Angled Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

1784-02 [Straight Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

Down Angled Jaw #1784-03

1784-03 [Down Angled Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Soudry Loose Body Grasper
Designed by Michael Soudry, MD

Designed to help with the removalof soft tissue loose bodies inarthroscopy and open procedures
PRODUCT NO:

1769

Overall Length: 9" (22,9 cm)
Shaft Length: 6" (15,2 cm)

6

INNOMED.NET   

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   FREE TRIAL ON MOST INSTRUMENTSSTANDARD LARGEPRODUCT NO’S:

OrthoVise™ Length: 10" (25,4 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3980-01 with Attachment Bolts (two sides & end)
without Slap Hammer
3981
without Attachment Bolts
without Slap Hammer
with End Attachment Nut that accepts aStandard Slap Hammer (#3925 or 3926)

3980

#3980
Side Attachment Bolts

Allows a Large OrthoVise™
Slap Hammer (#3950) to be
attached to the side of the device.

#3980-01

Available on Standard Large and Long Nose
Large OrthoVise™ with Attachment Bolts only.
(Cannot be added on later.)#3981

LONG NOSE LARGEPRODUCT NO’S:

OrthoVise™ Length: 12" (30,5 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3965-01 with Attachment Bolts (two sides & end)
without Slap Hammer

3965

#3965

OrthoVise™

#3965-01

U.S. Patent #D398,208

LONG NOSE LARGE BENT JAW
PRODUCT NO’S:

OrthoVise™ Length: 11.5" (29,2 cm)
with Attachment Nut (end)with Standard Slap Hammer (#3925)
3966-01 without Slap Hammer
with Attachment Nut (end) that accepts aStandard Slap Hammer (#3925 or 3926)

3966

#3966
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY#3966-01

Made of stainless steel
Models equipped with attachment bolts allow aslap hammer to be attached to the end, as wellas to either side of the large OrthoVise™ (except
Bent Jaw models), for greater adaptability
Bent Jaw models are not available with side
attachment bolts, but have an end attachment
nut to accept a Standard
Slap Hammer (#3925 or #3926)
A different size slap hammer is used for thelarge and small sizes of OrthoVise™
Slap Hammers are designed with a hammer
plate for the additionaluse of a mallet if desired

STANDARD SMALLPRODUCT NO’S:

OrthoVise™ Length: 8" (20,3 cm)
without Attachment Bolt
without Slap Hammer
3985-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3985-T with Attachment Bolt (end)without Slap Hammer

#3985

3985

#3985-01#3985-T

LONG NOSE SMALLPRODUCT NO’S:

OrthoVise™ Length: 9.5" (24,1 cm)
without Attachment Bolt
without Slap Hammer
3975-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3975-T with Attachment Bolt (end)without Slap Hammer

3975

SLAP HAMMERS#3975#3975-01#3975-T

PRODUCT NO’S:

3950 [Slap Hammer for Large OrthoVise]For use with 3965's, 3980's, 3981
Overall Length: 16.5" (41,9 cm)

For Large
OrthoViseFor use with: 3975's, 3985's
Overall Length: 8.75" (22,2 cm)

For Small
OrthoViseFor use with: 3966's
Overall Length: 16" (40,7 cm)

Standard
with 16" RodFor use with: 3966's
Overall Length: 16" (40,7 cm)

Easy Grip Standard
with 16" Rod3955 [Slap Hammer for Small OrthoVise]3925 [Standard Slap Hammer w/16" Rod]3926 [Easy Grip Slap Hammer w/16" Rod]

#3950
#3955
#3925
#3926

THREADED ADAPTERS
PRODUCT NO’S:

3980-02 [Small Adapter]#3980-02

Female/Female
Adapter Converts
from Male/MaleChanges Male End of a Slap Hammer to Female

Small Adapter

For use with 3965's, 3966's, 3980's, 3981

Threaded Large#3980-03

For use with: 3975's, 3985's

Threaded Small#3985-03

3980-03 [Threaded Adapting Screw – Large]

3985-03 [Threaded Adapting Screw – Small]

1.800.548.2362       JANUARY 2024   

Small Adapter allows a Standard Slap Hammer (#3925 or #3926)
to be used with any Large OrthoVise™ with Attachment Bolts

Threaded Adapting Screws can be used to append the corresponding size
OrthoVise™ with an Attachment Bolt for use with a Slap Hammer   KNEE INSTRUMENTS

7

Lotke Double Action
Cartilage Graspers
Designed by Paul Lotke, MD

Double action strength helps
to securely hold soft tissues
Standard #1710
Ratcheted #1715

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYPRODUCT NO’S:

1710 [Standard]

Overall Length: 7.5" (19,1 cm)

1715 [Ratcheted]

Angled to simulate the pinch forceps position.
Ferris-Smith tips effectively hold soft tissues orneedles. Powergrip avoids fatigue or excessive
forces on the surgeons thumbs.

Overall Length: 7.5" (19,1 cm)

Powers Modified Kocher Clamps
Designed by Mark Powers, MD

Heavier design allows for a firmer grasping of bone and soft tissues

PRODUCT NO’S:

1813 [Tapered Jaw]

Tapered Jaw

Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 5.2 mm x 4.1 mm#1813

1813-01 [Tapered Narrow Jaw]
Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 5.2 mm x 3 mm

Tapered Narrow Jaw#1813-01

1814 [Square Jaw]

Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 6.5 mm x 5 mm

Square Jaw#1814

Bhargava Knee Posterior Osteophyteand Anterior Hip Labral Grasper
Designed by Tarun Bhargava, MD

Very useful in helping to remove posterior osteophytesin knee surgery, and also to help remove the labrum
and soft tissues in anterior total hip surgery

8

PRODUCT NO:

1776

Overall Length: 12.5" (31,8 cm)
Shaft Length: 9" (22,9 cm)
Shaft Width: 7 mm
Jaw Width at End: 4 mm
Toothed Jaw Length: 14 mm

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Meniscal Clamp

Redesigned clamp is curved
for easier use, visualization,and tissue holding

PRODUCT NO:

1883

Overall Length: 7" (17,8 cm)
Teeth Length: .082" (2 mm)
Jaw Length: 1.5" (3,8 cm)

Bhargava Modified Meniscal Clamp
Designed by Tarun Bhargava, MD

Low-profile design helps facilitate grasping
the posterior portion of the meniscus

Improved bite when tension is placed on the meniscus.
Can also be used to help remove the fat pad and suprapatellar bursa.
PRODUCT NO:

1886

Overall Length: 7" (17,8 cm)
Teeth Length: 1 mm
Jaw Length: 1.125" (2,9 cm)

Delrin Insert Pliers

Designed to grasp an implant
for adjustment without
marring the implant surface
PRODUCT NO’S:

2025

Overall Length: 8" (20,3 cm)

2025-03 [Replacement Insert]
Includes top and bottom delrin jaws,
two screws and a hex wrench

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

9

Modified Rongeur with Pistol Grip Handle
Design modification by Morteza Meftah, MD and Ira Kirschenbaum, MD,of an original design by James T. Mazzara, MD.

A thin top cutter and deep lower cutter, withedges that are rounded off, allows the top
cutter to slide into a tight space— specificallythe acetabulum or the patella—while the pistol
grip helps lessen hand fatigue and slippage,
and allows for better visualization
PRODUCT NO:

1765

Jaw Bite Length: 18 mm
Jaw Bite Width: Tapered from 7 to 4.5 mm
Overall Length: 10" (25,4 cm)

Ortho Rongeur with Easy Grip Handle

Offset handle lessens hand fatigue and slippage,
and allows for better visualization
Offset handle gives better gripping power and helps reduce hand fatigue.
Finger grooves help to prevent hand slippage. The offset handle also
allows for better visualization. Available in three jaw bite sizes.

PRODUCT NO’S:

1780-01

Jaw Bite: 5 x 14 mm
Overall Length: 8.75" (22,2 cm)

1780-02

Jaw Bite: 7 x 16 mm
Overall Length: 8.75" (22,2 cm)

1780-03

Jaw Bite: 10 x 16 mm
Overall Length: 8.75" (22,2 cm)

5 x 14 mm#1780-01

7 x 16 mm#1780-02

10 x 16 mm#1780-03

Mazzara Rongeur
with Pistol Grip Handle
Designed by James T. Mazzara, MD

PRODUCT NO’S:

Pistol Grip handle lessens handfatigue and slippage, and allows
for better visualization

1765-01

Jaw Bite: 5 x 14 mm
Overall Length: 10" (25,4 cm)

1765-02

Jaw Bite: 7 x 16 mm
Overall Length: 10" (25,4 cm)

1765-03

Jaw Bite: 10 x 16 mm
Overall Length: 10" (25,4 cm)

10

INNOMED.NET   

5 x 14 mm#1765-01

7 x 16 mm#1765-02

10 x 16 mm#1765-03   FREE TRIAL ON MOST INSTRUMENTS

Mazzara Rongeur
for Small Bones
Designed by James T. Mazzara, MD

2 x 10 mm Jaw Bite

Designed for bone and soft tissue
removal in small joint surgery, thepistol grip handle lessens handfatigue and slippage, and allows
for better visualization#1765-04

4 x 10 mm Jaw Bite
#1765-05

Available in two jaw sizes.
PRODUCT NO’S:

1765-04

Jaw Bite: 2 x 10 mm
Overall Length: 7.25" (18,4 cm)

1765-05

Jaw Bite: 4 x 10 mm
Overall Length: 7.25" (18,4 cm)

PRODUCT NO’S:

1775-01 [Short Jaw]2 x 10 mm

Jaw widths
8 mm

Jaw Width: 8 mm
Overall Length: 9.25" (23,5 cm)

1775-02 [Medium Jaw]

Jaw Width: 5 mm
Overall Length: 9.25" (23,5 cm)

1775-03 [Long Jaw]

Jaw Width: 3 mm
Overall Length: 9.25" (23,5 cm)

4 x 10 mm

Hannum Tissue Grasper

5 mm

Designed by Scott Hannum, MD

3 mm

Teeth in jaw firmly holds
bone and tissue

MADE EXCLUSIVELYFOR INNOMED IN

Non-locking design can be easily gripped
while allowing greater pressure to be
applied. Available in three jaw sizes: short
jaw for holding bone, medium jaw forsmaller bones, and long jaw for tissue.

GERMANYLong Jaw Needle Nose Pliers

PRODUCT NO:

1833

Overall Length: 7" (17,8 cm)
Jaw Length: 2.25" (5,7 cm)
Jaw Width Tapered from: 8 mm to 1.5 mm
Jaw Height Tapered from: 12 mm to 2.5 mm

1.800.548.2362       JANUARY 2024   

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   KNEE INSTRUMENTS

11

Meftah PCL Protector
Designed by Morteza Meftah, MD

Designed to help protect theposterior cruciate ligament incruciate retaining total knee surgery
during the proximal tibial cut
The PCL Protector can be used efficiently right before thetibial cut. It is curved distally so that it can put over the PCLfrom the top/posterior side and with a few taps, the fanned
blade can get around the PCL and into the bone (not morethan 5 mm) and "cover" the PCL. The protector is left inplace until the tibial cut is made with a saw, which would hit
the protector instead of the PCL if it gets too close.
PRODUCT NO:

3221

Overall Length: 8" (20,3 cm)

McMaster Medullary Canal Aspirator

Designed by William McMaster, MD

Designed to aspirate the medullary canal prior toinsertion of the solid instrumentation alignmentrod to decrease the amount of semi-liquidmaterial present
Guide Wire
Canal TubePRODUCT NO’S:

8075

Overall Length: 19" (48,3 cm)

Also Available Individually:

8075-01 [Canal Tube]

Helps evacuate excess fat and marrowcontent from the medullary canal of a longbone, helping to reduce the pressure andforce created during insertion of a metal
rod into the canal, which can possibly
cause such materials to be emobilized
into the circuclation system (andeventually into the lungs) through open
venous structures.
The guide wire serves a dual purpose:
To help break up the medullary bone in
the proximal metaphysis to facilitate thepassage of the fenestrated rod, and after
the procedure to assist in cleaning andclearing the cannulated portion of the rod.
Also can be used on the tibial side if
an intramedullary guide system is used.
Can also be used during femoral rodding
procedures for fractures.

Overall Length: 18" (45,7 cm)

8075-02 [Guide Wire]

Overall Length: 19" (48,3 cm)

Tibiofemoral Offset Caliper
Designed by Adam Rosen, DO

A locking caliper designed to help accurately
measure the offset of the tibia from thesurface of the distal femur

PRODUCT NO:

5286

Overall Length: 3.75"– 6.25 (9,5 – 15,9 cm)
Overall Height: 2.75" (17,6 cm)
Width: 8 mm

12

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Wilson Condylar Gauge

Designed by Ralph Wilson, MD

Designed to measure the posterior femoral condyle after
the posterior cuts have been made in total knee arthroplasty
By measuring the depth of the residual condyle, the surgeon canresect excessive bone and measure the bone remaining to avoid
impingement of the condyle against the tibial component which could
impair knee flexion. The gauge is applied to the inferior or posterior
cut surface of the femoral condyle, and the back to front residual boneis measured and then removed as needed. Measures to 30 mm.
PRODUCT NO:

1194

Overall Length: 6" (15,2 cm)
Width: .568" (14,4 mm)

Ortho Caliper
Designed by Odell Woods
PRODUCT NO:

5285

Caliper: 0 to 12 cm
Leg Depth: 2" (5,1 cm)
Overall Length: 6" (15,2 cm)
Length expands to: 10.5" (26,7 cm)
Width: 8 mm

Mengato Depth Gauge

Designed by Richard Mengato, MD

Ring-handled design with 3 rings gives 3-point
grip for ease of holding and manipulation
Allows for superior gauge control and manipulation, toadvance, engage and maintain the hook on the distal cortex
by levering the probe against the bone hole and keeping
gentle tension on the hook.
PRODUCT NO:

1139

Overall Length – Contracted: 7.125" (18,1 cm)
Overall Length – Extended: 9.125" (23,2 cm)
Gauge: 0 to 50 mm

US Patent # 8,512,349

Depth Gauge

Designed for one-handed
use — helps to provide
measurement of thedepth/length of any bonehole for proper screw
length determination

PRODUCT NO:

8015

Overall Length: 7.625" (19,4 cm)
Scale: From 0 to 48 mm

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

13

Tibia AccuAngle

Can help toprovide a visual
aid in determining
if the centerline of the tibia
is 90 degrees
perpendicular to
the floor

Designed to be placed on the tibia
cutting block to check if the cut is level
Magnetic base helps to hold the AccuAngle in place
on a cutting block.

May also be used on
top of the tibia after
cut has been madeCutting
Block

Tibia
AccuAngle
Pin

Tibia
AccuAngle
CuttingBlock

A pin may be inserted
in the holes to provide
a visual reference ofthe cut's slope

PRODUCT NO:

1145

90˚

90˚

Dimensions:
2"x 3" (5,1 cm x 7,6 cm)

FLOOR

FLOOR

Trans-sulcus
Angle Guide
Line is drawn along
bottom of guide

A line is drawn down the deepest part of the trochlear sulcus(Whiteside line) with a marking pen or cautery. The post on
the guide is inserted into the hole in the femur made for an
intramedullary alignment guide. The trans-sulcus angle guide isthen rotated until the line on the guide lines up with the Whitesideline. A line is then drawn along the bottom of the guide.

Helps to establish
the trans-sulcus line

PRODUCT NO:

1160

Dimensions: 2.25" x .75" (5,7 cm x 1,9 cm)
Post Depth: 1.5" (3,8 cm)

Merchant Surgical Goniometer
Designed by Alan Merchant, MD

Designed to help assess frontal plane
limb alignment or measure the Q angle

Collapsible and steam sterilizable.

The extended length can reach from the center of the kneeto the femoral head or the anterior superior iliac spine.
The collapsible stainless steel device is autoclavable.
PRODUCT NO:

2029

Overall Length: 41" Fully Extended (104,2 cm)      22.5" Folded in Half (57,2 cm)      12" Fully Collapsed (30,5 cm)

14

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Articulated Measuring Device
with Ruler
Designed by Vincent Y. Ng, MD

A highly precise (within 1 mm) device
designed for measuring distances
between two points — can be used even
if there are intervening structures like
soft tissue or bone, and in situations
where a straight ruler will not work
Example of use include measuring limb length in total hip
arthroplasty, confirming length in megaprosthetic knee
replacements, and assessing dimensions of allografts.
PRODUCT NO’S:

2026-00 [Measuring Device with Ruler]
Set Includes/ Available Separately:2026-01 [Measuring Device Only]

Overall Length (unfolded): 15.25" (38,8 cm)
Dimensions Triangle Folded: 4" x 4.25" (10,2 x 10,8 cm)

2026-02 [Ruler Only]

Overall Length: 9" (22,9 cm)
Width: .79" (2 cm)

Use punch to
remove tibial
bone plug, then...

...use tamp to insert
plug as autograft for the
femoral intramedullaryalignment hole

Goytia Osteotome Punch
Tamp Assembly
Designed by Robin Goytia, MD

Designed for removing a tibial boneplug to use as autograft for the
femoral intramedullaryalignment hole in total
knee replacement

PRODUCT NO’S:

5339-00 [Punch & Tamp Set]
Set Includes / Available Individually:

Overall Length: 7.75" (19,7 cm)
Outside Diameter: 16 mm
Inside Diameter: 13.7 mm

5339-02 [Tamp]

Overall Length: 7.75" (19,7 cm)
Diameter: 12.3 mm

1.800.548.2362       JANUARY 2024   

Punch #5339-01

5339-01 [Osteotome Punch]   KNEE INSTRUMENTS

Tamp #5339-0215

2° Right Rasp #6906-03

0° Flat Rasp #6906-02

2° Left Rasp #6906-04

Grant TKA Anatomic
Bone File Set
Designed by Richard E. Grant, MD

A bone rasp and plumb rod
set designed for TKA tibial
cut surface preparation
PRODUCT NO’S:

Plumb Rod6906-00 [Set]#6906-01

Set Includes/ Available Separately:6906-01 [Plumb Rod]Overall Length: 14" (35,6 cm)

6906-02 [0° (Flat) Rasp]

Overall Length: 6.375" (16,2 cm)
Rasp Platform Length: 1.7" (4,3 cm)
Rasp Platform Width: 2.7" (6,9 cm)

6906-03 [2° Right Rasp]

Overall Length: 6.375" (16,2 cm)
Rasp Platform Length: 1.7" (4,3 cm)
Rasp Platform Width: 2.7" (6,9 cm)

6906-04 [2° Left Rasp]

Overall Length: 6.375" (16,2 cm)
Rasp Platform Length: 1.7" (4,3 cm)
Rasp Platform Width: 2.7" (6,9 cm)
Patent Pending

Plumb rod fits into the handle of each bone rasp: 0°, 2° Left, and 2° Right.

Colwell TKA 5° Tibial Rasp Assembly

PRODUCT NO’S:

6900-00 [Complete Assembly]
Overall Length: 15" (38,1 cm)
Set includes:

Designed by Clifford W. Colwell Jr., MD

A tibial planing tool with a universal design to help
improve tibial cut alignment and flatness by smoothing out
imperfections intraoperatively, helping to ensure the tibial
bone surface is cut correctly in coronal and sagittal planes

6901-01 [Rasp Handle]

Overall Length: 6.283" (16 cm)
Handle Hength: 3.625" (9,2 cm)

6901-02 [Rasp Plate]

Plate Width: 2.65" (7,7 cm)
Plate Depth: 1.75" (4,3 cm)

6902 [T-Handle Canal Rod]

After the planer rasp handle/plate unit is threaded onto the intramedullaryrod, the handle is moved back and forth through an arc while the cutting
surface of the planer is held against the tibial bone, to realign the cut and to
remove any imperfections. For use with any primary or revision knee system
when an intramedullary cutting guide is being used.

16

Overall Length: 15" (38,1 cm)
T-Handle Width: 4" (10,1 cm)

6903 [Handle Grip]

Overall Length: 4" (10,1 cm)

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Tibial Impactor
Design modified by Atul F. Kamath, MD

Assists in MIS unicompartmental
cemented tibial tray impaction, andcan also be helpful for impaction ofother components such as ankle
PRODUCT NO’S:

1129

Dimensions: 7" x 4" (17,8 cm x 10,2 cm)
Delrin Impactor Pad: 1" x .625" (2,5 cm x 1,6 cm)

Replacement Part:1129-02 [Replacement Pad Only]

Boynton Punch
Designed by L. Boynton, MD

Helpful in removing trial, femoral
and revision total knee components
The flange end fits onto the flange of a femoral
knee component or trial.

Offset #5120-02
PRODUCT NO’S:

Overall Length: 11.75" (29,8 cm)
Shaft Diameter: 9,5 mm

Stainless

Impactor Sizes

Standard #5120-01

5120-02 [Offset]

5120-01 [Standard]

DelrinOverall Length: 11.75" (29,8 cm)
Shaft Diameter: 9,5 mm
Punch End Offset: 60 mm

Impactor Sizes

Modular
Impactor
Set
Makes multiple impactor heads
easily visible and available

Designed to have available to the operating
surgeon multiple types of impactors utilizing one
handle. The rack uses less space and allows thesurgeon to quickly see the designs available. The
impactors are supplied with stainless steel tips forbone and delrin tips which can be used against
an implant for slight placement adjustments.

PRODUCT NO:

5370 [Complete Set]
Included In Set / Also Available Individually:

5370-01 [Rectangular Tip 11 mm x 4 mm Steel]
5370-02 [Oval Tip 13 mm x 8 mm Steel]
5370-03 [Crescent Tip 12 mm x 5 mm Steel]
5370-04 [Square Tip 9 mm x 9 mm Steel]
5370-05 [Round Tip 15 mm Steel]
5370-06 [Round Tip 12 mm Steel]
5370-07 [Round Tip 9 mm Steel]
5370-19 [Set Base] Base Diameter: 3.5" (8,9 cm)
5370-D1 [Rectangular Tip 11 mm 4 mm Delrin]
5370-D2 [Oval Tip 13 mm x 8 mm Delrin]
5370-D3 [Crescent Tip 12 mm x 5 mm Delrin]
5370-H [Modular Handle]
Overall Length: 8" (20,3 cm)
Grip Length: 4.5" (11,4 cm)

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

17

Bacastow Tibial Plateau Elevators

Designed by David Bacastow, MD

Designed to help with indirect reduction of a depressedtibial plateau fracture, and can be used with arthropscopic
visualization and percutaneous fixation

PRODUCT NO'S:

5297 [Starter Elevator]

Finish 10.4 mm

Overall Length: 11" (27,9 cm)
Tamp Diameter: 4,7 mm#5298

Starter 4.7 mm

5298 [Finish Elevator]#5297

Overall Length: 11" (27,9 cm)
Tamp Diameter:10,4 mmMalleable Bone Tamps

Modified by Serge Kaska, MD

The large tamp is designed to help elevate a depressed tibial plateau
fracture, while the small tamp can help elevate a depressedtibial plafond and smaller tibial plateau fractures
PRODUCT NO’S:

5296 [Large]

Overall Length: 14" (35,6 cm)
Shaft Length: 9.5" (24,1 cm)
Impactor Diameter: 12.5 mm

5296-01 [Small]

Overall Length: 9.5" (24,1 cm)
Shaft Length: 6" (15,2 cm)
Impactor Diameter: 10 mm

5296-02 [Extra Small]

Overall Length: 11.4" (29 cm)
Shaft Length: 5.9" (15 cm)
Impactor Diameter: 6.5 mm

Large 12.5 mm#5296

Small 10 mm#5296-01

Extra Small 6.5 mm#5296-02

Malleable shaft can be contoured for different angles

Sandman Curved Bone Punch
Designed by Geoffrey A. Sandman, MD

Designed to help elevate a depressedtibial plateau fracture

PRODUCT NO:

5305

Overall Length: 14" (35,6 cm)
Shaft Length: 9.5" (24,1 cm)
Impactor Diameter: 12,5 mm (.5")

18

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Faillace Bone Impact/Graft Forceps
Design modification by John J. Faillace, MD, FAAOSLong vertical grooves at the tip are designed to deliver graft
into a small space, where a freer elevator can be used topush the graft down into the space, then the closed
flat end can be used to tamp down the graft
PRODUCT NO:

MADE EXCLUSIVELYFOR INNOMED IN

5011

Overall Length: 5" (12,7 cm)
Tip Diameter When Closed: 3,2 mm

Universal Bone
Grafting/Impacting
Forceps

GERMANY

Bone graft can be grasped, placed& impacted without changing
hands or instruments

Designed by J. A. Amis, MD

The forceps are designed with graspingends for delivery of bone graft. When thegraft is in place, the forceps are closed,
which forms the ends into an impacting
punch. A striking platform is attached tothe end of the forceps for tapping andtamping the graft. Four end diameters
are available in two lengths.
PRODUCT NO’S:
Short: 6" (15,2 cm) Length

1/8"
3/16"
1/4"
5/16"
(3,2 mm) (4,8 mm) (6,3 mm)(8 mm)
Diameter ends at actual size (closed forceps)

5010-01 1/8" (3,2 mm) Diameter End
5010-02 3/16" (4,8 mm) Diameter End
5010-03 1/4" (6,3 mm) Diameter End
5010-04 5/16" (8 mm) Diameter End

Long: 10" (25,4 cm) Length

5050-01 1/8" (3,2 mm) Diameter End
5050-02 3/16" (4,8 mm) Diameter End
5050-03 1/4" (6,3 mm) Diameter End
5050-04 5/16" (8 mm) Diameter End

9 mm Round#5337

12 mm Round#5336

15 mm Round#5335

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

9 mm Square
#5334

Ortho
Impactors

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

When the forceps areclosed, they form into
an impacting punch

2 mm Tapered#5333

12 x 7 mm Rectangle 11 x 4 mm Rectangle
#5332

#5331

PRODUCT NO’S:

Overall Length: 9" (22,9 cm)
Shaft Diameter: 9 mm

5331 [11 x 4 mm Rectangle]
5332 [12 x 7 mm Rectangle]
5333 [12 mm Tapered]5334 [9 mm Square]5335 [15 mm Round]5336 [12 mm Round]5337 [9 mm Round]19

Redler Bone Clamp with Wire Guide

Designed by M.R. Redler, MD

Designed to hold bony fragments inplace for placement of guide wires
Can be used for:Placement of guide wires during the open reduction
and internal fixation of a patella fracture
Placement of pins across distal radius fractures or
across carpal bones
Arthroscopically assisted fixation in the wrist
Fracture fragments about the elbow

Pins Up To .062" / 1.6 mm#1885-62

Pins Up To .045" / 1.1 mm#1885-45

PRODUCT NO’S:

1885-45

For Pins up to .045" (1,1 mm)
Overall Length: 9.5" (24,1 cm)
Jaw opens to: 3.5" (8,9 cm)

1885-62

For Pins up to .062" (1,6 mm)
Overall Length: 9.5" (24,1 cm)
Jaw opens to: 3.5" (8,9 cm)

Scott Patella Resection Guide/Clamp
Designed by James Scott, MD

Helps move the tendons anteriorly, giving
the surgeon a good method of holding thepatella stable for resection

47 mm
54 mm

Can be used as a holding device, or as a
guide if the surgeon uses the tendon insertion
to the patella as level for resection.
PRODUCT NO:

1164

Overall Length: 10" (25,4 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Patella Grasping Forceps
Designed by S. David Stulberg, MD

Bent handle on forceps helps thesurgeon to evert the patella during
minimally invasive knee surgery

PRODUCT NO:

4250

Overall Length: 6.75" (17,1 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Normally two forceps are used (sold individually)

Patella Cover Plate

Designed by S. David Stulberg, MD

Protects the cut surface of the patelladuring minimally invasive knee surgery
Flat topside, with three small spikes underneath

Sharp spikes help hold the plates in place. Lessens the chanceof weakening the patella, as pre-drilling is not necessary.
PRODUCT NO’S:

4230-00 [Set of 4 Sizes]
4230-01 [Small] 35 mm x 31 mm
4230-02 [Medium] 36 mm x 32 mm
4230-03 [Large] 37 mm x 33 mm
4230-04 [Extra Large] 38 mm x 34 mm

20

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Desai Curette Osteotomes

Designed to remove bone and cartilage, helpful for preparing joint
surfaces for fusion, allowing easy removal of osteophytes andcartilage without having to switch instruments

The osteotome portion also can be used to "feather" the subchondral surface toexpose bleeding bone. It is also useful in instances of obtaining autograft, as it
can be used to create a bone window and then remove cancellous bone.
Designed by Sarang Desai, DO

5241 [5 mm]

Cup Sizes:

5 x 6 mm Cup

PRODUCT NO’S:#5241

5242 [8 mm]

Overall Length: 8.25" (21 cm)
Overall Length: 8.25" (21 cm)
Handle Length: 4.25" (10,8 mm)
Handle Length: 4.25" (10,8 mm)
Cup: 5 x 6 mm
Cup: 8 x 10 mm
Osteotome Width: 3.5 mm
Osteotome Width: 6.5 mm
Osteotome Length: 3.5 mm from
Osteotome Length: 3 mm from        edge of cup         edge of cup

5 x 6 mm

8 x 10 mm Cup#5242

8 x 10 mm

Wagner Osteotome Handle
Designed by Russell Wagner, MD

Handle is designed for easier gripping, rotational
control, and use with a mallet with a standard
1/4” Lambotte osteotome
PRODUCT NO'S:

5348 [Handle Only]

Overall Length: 5.5" (14 cm)

5348-01 [1/4" Osteotome Only]
Overall Length: 8.875" (22,5 cm)

Nordt Precision Micro Fracture Set

20° Bent Awl #8025-01

Designed by William E. Nordt, III, MD
PRODUCT NO’S:

40° Bent Awl #8025-02

8025-00 [Complete Set]
Also available individually:

8025-01 [20° Bent Awl]
Overall Length: 10" (25,4 cm)

Angled Osteotome #8025-03

8025-02 [40° Bent Awl]
Overall Length: 10" (25,4 cm)

8025-03 [Angled Osteotome]
Overall Length: 10.875" (27,6 cm)

8025-04 [Bent Stirrup Scraper]

Bent Stirrup Scraper #8025-04

Overall Length: 10.125" (25,7 cm)

Ultra hard titanium
nitride coating helps toextend life by increasing
surface hardness, prolonging
sharpness, and resistingchemicals and corrosion.

8025-05 [Tri-Tip Awl]

Precise microfracture points
Helps create sharp cartilage shoulders

Overall Length: 10" (25,4 cm)

8025-CASE [Case]

Tri-Tip Awl #8025-05

Whelan Double-Ended Suture Wire Passer

Passer guide and malleable passer designed
to pass suture wires around a bone

Designed by Edward J. Whelan, III, MD
PRODUCT NO’S:

8300-00 [Set]
Also available individually:

8300-01 [Passer Guide]

Overall Length: 8.125" (20,6 cm)
Oustide Width: 9 mm
Inside Groove Width: 6,5 mm

8300-02 [Passer]

Overall Length: 7.5" (19,1 cm)
Width: 4,6 mm

1025 [Sterilization Case]

PasserPasser Guide
Set includes
PasserGuide, twoPassers, anda sterilizationcase.

#8300-01#8300-02

The passer guide is placed around the bone, andthe thin malleable passer is inserted at the handle end
and follows the grooved passer around. The suture wire (up to 18 gauge)
is attached to the keyholed end of the passer, which can then be reversed out ofthe passer, which can then be reversed out of the passer, drawing the suture wire around the bone.

1.800.548.2362       JANUARY 2024   

Insert passer
into guideto pass aroundthe bone   KNEE INSTRUMENTS

Attach suture
wire, then draw
the passer/
suture wire backaround the
bone

21

Durham Curved Osteotome
Designed by Alfred A. Durham, MD

Increased angle useful for posterior
osteophytes of the femoral condyleand the humeral head, as well asanterior acetabular osteophytes
PRODUCT NO:

4950

Overall Length: 9" (22,9 cm)
Handle Length: 5" (12,7 cm)
Osteotome Width: .625" (1,6 cm)

Cobb Elevators

Two Sizes Available With
or Without Teeth

PRODUCT NO’S:
WITH TEETH

Ultra hard titanium nitride
coating helps to extend blade
life by increasing surface
hardness, prolonging sharpness,and resisting chemicals andcorrosion.

3432 [1/2" with Teeth]

1/2" with Teeth #3432
1/2" without Teeth #3436

Overall Length: 11" (27,9 cm)
Blade Width: 1/2" (13 mm)

3434 [1" with Teeth]

Overall Length: 11" (27,9 cm)
Blade Width: 1" (25,4 mm)

WITHOUT TEETH

3436 [1/2” without Teeth]

Available withor without teeth

Overall Length: 11” (27,9 cm)
Blade Width: 1/2” (13 mm)

1" with Teeth #3434
1" without Teeth #3438

3438 [1” without Teeth]
Overall Length: 11” (27,9 cm)
Blade Width: 1” (25,4 mm)

Bradley Periosteal Elevator
Designed by Gary W. Bradley, MD

Ultra hard titanium nitride coating helps to extend
blade life by increasing surface hardness, prolonging
sharpness, and resisting chemicals and corrosion.
PRODUCT NO’S:

4719 [1/2"]

Overall Length: 11" (27,9 cm)
Blade Width: .5" (13 mm)

1/2" #4719

4720 [3/4"]

Overall Length: 11" (27,9 cm)
Blade Width: .75" (19 mm)

3/4" #4720

Periosteal Elevator

Designed for better control
Designed with a curved end for easieruse, and sharper sides for ease ofelevating and stripping. The handle isdesigned for better control.

22

PRODUCT NO’S:

3450 [Curved]

Overall Length: 7.5"
Handle Length: 4.5" (11,4 cm)
Blade Size: 16 x 13 mm

Curved #3450

3455 [Straight]

Overall Length: 7.75"
Handle Length: 4.5" (11,4 cm)
Blade Size: 19 x 14 mm

Straight #3455

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Modified Lambotte Osteotomes

Designed with a striking platform, plus a cross-bar hole tohelp control rotational stability and assist with removal
Six (6) sizes available, from 1/4" to 1-1/2" in 1/4" increments. Cross-bar
and case included in complete set. Two smallest sizes have an 1/8" holein which an 1/8" pin can be used as a cross bar (not included).
PRODUCT NO’S:* MADE EXCLUSIVELY

5350-00 [Set w/Case]

FOR INNOMED IN

GERMANY

Also Available Individually:

5350-25* [1/4"]

Overall Length: 9" (22,9 cm)
Osteotome Width: .25" (6,4 mm)

5350-125 [1-1/4"]

Overall Length: 9" (22,9 cm)
Osteotome Width: .5" (12,7 mm)

5350-150 [1-1/2"]

Overall Length: 9" (22,9 cm)
Osteotome Width: .75" (19 mm)

5350-CASE [Case]

Overall Length: 9" (22,9 cm)
Osteotome Width: 1" (25,4 mm)

5350-CB [Cross Bar]5350-50* [1/2"]
5350-75 [3/4"]
5350-100 [1"]Overall Length: 9" (22,9 cm)
Osteotome Width: 1.25" (31,8 mm)
Overall Length: 9" (22,9 cm)
Osteotome Width: 1.5" (38,1 mm)
Dimensions: 12.25" x 11.25" x 1"
(31,1 x 28,6 x 2,5 cm)
Overall Length: 4.375" (11,1 cm)

Offset Osteotomes

Designed to remove osteophytes
from the posterior femoral
condyles during knee arthroplasty

Wide Offset Osteotome
Designed by Paul Lotke, MD & Adam Rosen, DO
PRODUCT NO:

4920

Blade Width: 18,5 mm
Overall Length: 9" (22,9 cm)

Wide Offset #4920

Lotke Offset Osteotome

Designed by Paul Lotke, MD
PRODUCT NO:

4935

Blade Width: 13 mm
Overall Length: 9" (22,9 cm)

Lotke Offset #4935

Dennis Offset Osteotome
Designed by Douglas Dennis, MD & Paul Lotke, MD
PRODUCT NO:

4935-W

Blade Width: 18,5 mm
Overall Length: 9" (22,9 cm)

Dennis Offset #4935-W

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

23

UKA Tibial Bone Fenestrator
Designed by Todd Borus, MD

Designed for improving cement
penetration during UKA

With the minimal bone resection of modern UKA systems,
often the tibial and femoral surfaces can remain quite
sclerotic after bone cuts are performed. Instrument isdesigned to allow fenestration of the entire bone surface,helping to promote optimal cement interdigitation during UKA.
PRODUCT NO:

8026

Overall Length: 8.875 (22,5 cm)
Handle Length: 4.5" (11,4 cm)

Woolley Tibia Punch

Designed by D. Woolley, MD

Designed to impact cancellous bone to help improve bone/cement interface
Designed to impact cancellous bone in the subchondral weight bearing region of the tibia. This helps to improvethe mechanical interlock in the cancellous bone/cement interface. The sharp tips can be used on
normal and dense cancellous bone, and they can also be used when a significant
deformity has been encountered resulting in sclerotic bone.

PRODUCT NO:

5140

Prong Depth: 5,5 mm
Overall Length: 7" (17,8 cm)
Shaft Diameter: 13 mm

Lombardi Tibia Cement Preparation Drill

Designed by Adolph Lombardi, MD

Designed to drill cancellous bone to help improve bone/cement interface
Designed to drill cancellous bone in the subchondral weight bearing regionof the tibia. This helps to improve the mechanical interlock in the cancellous
bone/cement interface. Connects with a Zimmer Hall Quick-connect.

Quick-connect for use with a driver.
PRODUCT NO’S:

1112 [Lombardi Tibia Cement Preparation Drill]
Drill Diameter: 2.7 mm
Drill Length: 3 mm
Overall Length: 4.75"

8248 [Fixed Driver with Zimmer Hall Quick-connect]
Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

24

INNOMED.NET   

Fixed Driverwith Zimmer Hall
Quick-connect   FREE TRIAL ON MOST INSTRUMENTS

Kodkani Tissue Elevator Suture/Graft Passer
Designed by Pranjal Kodkani, MD

Designed for MPFL reconstruction basket
weave technique, and helpful for mini- open
ligament reconstruction surgeries
for graft passage
No Slot #1114
PRODUCT NO’S:

1114 [No Slot]

Overall Length: 9.75" (24,8 cm)
Handle Length: 4.25" (10,8 cm)
Suture Hole: 2,5 mm x 13 mm

1114-01 [With Slot]

Overall Length: 9.75" (24,8 cm)
Handle Length: 4.25" (10,8 cm)
Suture Hole: 2,5 mm x 13 mm

With Slot #1114-01

Can also be used for:
Periosteum/soft tissue elevator or freer
Percutaneous passage of tendon/ligament graft/suture
Stripping tendon grafts off muscle
General orthopedics – repiosteum elevator and spike

Wilson Double Scalpel Handle

Advantage of the open slot:
Convenient feeding and removal of sutures from slot
Feeding of multiple thick sutures & sutures with knots
Engaging and shuttling grafts with short suture loop ends

Designed by Ralph Wilson, MD

Designed to help make a predictable incision in thepatellar tendon when harvesting ACL graft material
The blade offset is 10 millimeters. The tendon graft is harvested
from the patella and tibial tubercle including the patellar tendon.
Uses scalpel blades that fit a #3 handle size.
PRODUCT NO:

8207

Overall Length: 5.75" (14,6 cm)

Scalpel blades not included.

Scalpel blades not included.

Seymour ACL Graft Advancer
Designed by Scott Seymour, MD

Designed to facilitate the passage
and tensioning of an ACL graft into
the femoral and tibial tunnels
A loop is tied in the prepared graft's passing sutures
and the device is used to pull the graft into thetunnels, then to tension the fixation.
PRODUCT NO:

1117

Overall Length: 4.35" (11,1 cm)
Handle Width: 4" (10,2 cm)
Hook Width: 19,5 mm Outside, 13,5 mm Inside
Hook Depth: 25 mm
Hook Diameter: 3 mm

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

25

Pin Inserter

Used for 1/8" (3,2 mm) diameter pin insertion

Designed to hold onto a 1/8" (3,2 mm) diameter pin while it isbeing inserted into a cutting block during total knee surgery
or other applications where a 1/8" (3,2 mm) diameterpin is used. The pin inserter holds the pin tightly,
yet releases it easily after insertion. It may
be used with round or
triangular end pins.

PRODUCT NO:

4020

Overall Length: 5" (12,7 cm)

Pin Inserter/Extractor

Helps provide better leverage, stability
and control when inserting/extracting pins

Completely cannulated, allowing use on long pins,
where the instrument can be next to the bone orskin for stability and control. The grasping end
is contoured to not block the surgeon’s fieldof view. The handle is shaped so not to slide in
the surgeon’s hand and for better leverage. May
also be used to pull a drain needle from thesurgical site. The design helps to protect operating
personnel from the sharp tip of the needle.
The slap hammer is designed to be screwed
into a threaded pin inserter/extractor to help in
removing pins in hard bone.

PRODUCT NO’S:

3020 [For 1/8" (3,2 mm) Pins]
3020-T-00 [For 1/8" (3,2 mm) Pins, with
Slaphammer and Case]
3020-T [For 1/8" (3,2 mm) Pins, Threaded
to Accept slap hammer]
3040 [Slap Hammer]
Thread: 5/16"x 181015 [Sterilization Case]

Pin Driver and Threaded Bone Pins
85 mm Threaded Pin #1287

Pin Driver with Zimmer Hall Quick-connect #1206 Quick-connect version for use with a driver.

65 mm Threaded Pin #1290
55 mm Threaded Pin #1297
PRODUCT NO’S:
1/8" (3,2 mm) Pins – Packages of 10:

1287 [85 mm Threaded Bone Pin]
1290 [65 mm Threaded Bone Pin]
1297 [55 mm Threaded Bone Pin with Collar]

Pin Driver #1205

Fixed Driver with
Zimmer Hall Quick-connect

PRODUCT NO’S:

1206 [Pin Driver w/Zimmer Hall Quick-connect]
Overall Length: 5" (12,7 cm)

1205 [Pin Driver]

Overall Length: 3.75" (9,5 cm)

8248 [Fixed Driver with Zimmer Hall Quick-connect]
Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

Shouldered Bone Pins
PRODUCT NO’S:
Packages of 10:

1/8" / 3.2 mm #1270

1270 [1/8"]

Diameter: 3,2 mm (.125")
Overall Length: 70 mm
Shoulder-to-tip: 45 mm

1271 [1/16"]

1/16" / 1.6 mm #1271

Diameter: 1,6 mm (.062")
Overall Length: 70 mm
Shoulder-to-tip: 45 mm

26

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS#8248

Stanton Straight
Pin Removal Pliers
Designed by John Stanton, MD
PRODUCT NO:

1893

Overall Length: 6.375" (16,2 cm)
Jaw Length: 1.62 (4,1 cm)
Instrument Width: 1 cm

Ring Curettes – Straight Shaft

Ring Curettes – Bent Shaft

3 mm Straight #5150

3 mm Bent #5156

6 mm Straight #5152

6 mm Bent #5157

8 mm Straight #5154

8 mm Bent #5158

PRODUCT NO’S:
Straight Shaft

Overall Length: 8.75" (22,2 cm)

5150 [3 mm Straight]
Ring Diameter: 3 mm

PRODUCT NO’S:
Bent Shaft

MADE FOR INNOMED IN

GERMANY

5152 [6 mm Straight]

Ring Diameter: 6 mm

5154 [8 mm Straight]
Ring Diameter: 8 mm

Overall Length: 8.625" (21,9 cm)

5156 [3 mm Bent]

Ring Diameter: 3 mm

Sarraf Toothed Curettes

MADE FOR INNOMED IN

GERMANY

5157 [6 mm Bent]

Ring Diameter: 6 mm

5158 [8 mm Bent]

Ring Diameter: 8 mm

Forward, straight, and reverse bent toothed curettes
designed to aid in aid in all types of joint arthroplasty
surgery, especially in scraping any articularchondral islands within the acetabulum
during THA preparation

Designed by Khaled Sarraf, MD
PRODUCT NO’S:

5174-00 [Set]
Set Includes/ Available Separately:

5174-F [Forward Toothed Curette]
Overall Length: 11.5" (29,2 cm)
Handle Length: 5.5" (14 cm)
Curette Cup : 8 mm X 12 mm
Angled Down: 30°

5174-R [Reverse Toothed Curette]
Overall Length: 11.5" (29,2 cm)
Handle Length: 5.5" (14 cm)
Curette Cup : 8 mm X 12 mm
Angled Up: 30°

5174-S [Straight Toothed Curette]
Overall Length: 11.5" (29,2 cm)
Handle Length: 5.5" (14 cm)
Curette Cup : 8 mm X 12 mm

Forward Toothed
Can also be used for the femoral
canal in cemented and uncemented THA
Valuable aid in revision arthroplasty (hip, knee,shoulder and ankle) for cement curettage
Useful tool in hip and knee primary arthroplasty as well
as shoulder, elbow and ankle arthroplasty procedures
Ultra hard titanium nitride coating helps to extend life by increasing surface
hardness, prolonging sharpness, and resisting chemicals and corrosion.

#5174-F

Reverse Toothed#5174-R

Straight Toothed#5174-S

Chandran Bent Serrated Curette
Designed by Rama E. Chandran, MD

PRODUCT NO:

5171

Overall Length: 11.75" (29,8 cm)
Handle Length: 5.5" (14 cm)
Cup Size: 7 mm X 12 mm

Serrated design allows for easier removal of cancellous
bone in the proximal femur in total joint arthroplasty
Ultra hard titanium nitride coating helps to extend life by increasing surface
hardness, prolonging sharpness, and resisting chemicals and corrosion.

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

27

Engh
Cement Scrapers

Left#4920-02

Right#4920-01

Designed by Gerard A. Engh, MD

Right and left design used toscrape cement from aroundand behind knee implants
PRODUCT NO’S:

4920-01 [Right]

Scraper Head: 5 mm x 9 mm
Overall Length: 8.5" (21,6 cm)

4920-02 [Left]

Scraper Head: 5 mm x 9 mm
Overall Length: 8.5" (21,6 cm)

Scott Uni & Total Knee Cement Removing Curette
Designed by Richard D. Scott, MD

Sized, shaped and angled 90° to help with retrieval of posteriorly extruded cement
behind the tibial component in both total and unicompartmental knee arthroplasty
Ultra hard titanium nitride coating helps to extend curette life by increasing surface hardness,prolonging sharpness, and resisting chemicals and corrosion, while helping toeliminate metal transfer and protect the implant surface.

PRODUCT NO:

4247

Overall Length: 9.625" (24,4 cm)
Overall Length: 5.25" (13,3 cm)
Cup Size: 4/0

Curved Cement Osteotome

Helps remove cement around the back of the tibia base, and useful inthe femoral notch during removal of a knee femoral component
Designed to be inserted around the back of the tibia base to remove cement. The curve is congruent
with most tibia bases. During revision knee surgery, can be used to help separate theprosthesis/bone or prosthesis/cement interface. The curve of the osteotome
allows it to be used in the femoral notch of a femoral component. The
osteotome is nitrate coated to help protect the implant surface.
PRODUCT NO:

5220

Overall Length: 6.75" (17,1 cm)
Handle Length: 3" (7,6 cm)
Blade Width: 6,8 mm

Cement Remover

Helps remove unhardened cement aroundfemoral and tibial knee components
Designed with a sharper face to help remove
unhardened cement around femoral andtibial knee components. The removeris nitrate coated to help protect
implant surfaces.

28

INNOMED.NET   PRODUCT NO:

5230

Overall Length: 7.25" (18,4 cm)
Handle Length: 5" (12,7 cm)
Blade Width: 5 mm   FREE TRIAL ON MOST INSTRUMENTS

Bacastow Femoral Cement Osteotome
Designed by David Bacastow, MD

Uniquely shaped osteotome designed
to help trim away cement from arounda femoral knee component
PRODUCT NO:

5234

Overall Length: 9.25" (23,5 cm)
Width: 6,5 mm
Tongue Length: 7 mm

Bozeman Cement Trimmer
Designed by Daniel M. Gannon, MD

The tool has a blunt blade tip on one end to help withseparation of the trimmed cement. The angled curette end
helps gather the trimmed cement. The thin shank and angled
curette can reach into tight spaces such as the back of theimplants to remove excess cement. The ends are titanium
nitrite coated to help eliminate metal transfer.

Combines the two most common
cement trimming tools into one

PRODUCT NO:

5245

Overall Length: 8.5" (21,6 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Seachris Delrin Cement Scraper
Designed by Timothy Seachris

Reusable delrin scraper is designed
to help remove cement around aknee or hip prosthesis
PRODUCT NO:

5218

Overall Length: 5" (12,7 cm)
Thickness: 1/8" (3 mm)

Robb Cement Curette
Designed by William Robb, MD

Designed to help remove cement
around a hip or knee prosthesis
Made of Delrin
PRODUCT NO:

5635

Overall Length: 8" (20,3 cm)
Freer End: 5 mm
Cup End: 10 mm

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

29

Sarraf TiN Coated
Cement Removal
Forceps

Straight #5039
Bent #5041

Designed by Khaled M. Sarraf, MD
PRODUCT NO’S:

5039 [Straight]

Overall Length: 6" (15,2 cm)

5041 [Bent]

Overall Length: 6.125” (15,6 cm)

Ultra hard titanium nitride coating helps to extend forceps life by increasing surface hardness, prolonging sharpness,and resisting chemicals and corrosion, while helping to eliminate metal transfer and protect the implant surface.

The curved semicircular tip is congruent
to most tibial plates and femoral condylar
implants, helping to facilitate removal of excess
cement, especially at the tight posterior aspect
The small scoop-end tip assists in excising
unset cement
Ultra hard titanium nitride coating helps
to extend curette life by increasing surface
hardness, prolonging sharpness, and resistingchemicals and corrosion, while helping toeliminate metal transfer and protect the
implant surface

Sarraf Cement Trimmer
Designed by Khaled M. Sarraf, MD

Two-in-one instrument designed for cement
removal during arthroplasty surgery

Sarraf Spearhead Cement Exciser
Designed by Khaled M. Sarraf, MD

PRODUCT NO:

Two-in-one instrument designed for cement
removal during arthroplasty surgery

5211

Overall Length: 7.75" (19,7 cm)

PRODUCT NO:

5212

Overall Length: 7.75" (19,7 cm)

The curved semicircular tip is congruent
to most tibial plates and femoral
condylar implants, helping to facilitate
removal of excess cement, especially at
the tight posterior aspect
The spearhead tip assists in excisingand shaping the unset cement
Ultra hard titanium nitride coating helps
to extend curette life by increasing
surface hardness, prolonging sharpness,and resisting chemicals and corrosion,
while helping to eliminate metal transfer
and protect the implant surface

Cement Packer & Trimmer
Designed by Harlan C. Amstutz, MD

PRODUCT NO:

4995

MADE FOR INNOMED IN

GERMANY

Overall Length: 9.75" (24,8 cm)

30

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Gelbke Freer Cement Trimmer/Nerve Hook with TiN Coating

Designed by Martin K. Gelbke, MD

Designed to facilitate cement removal during total
and partial knee replacement

Consists of a freer elevator on one end and a nerve hook on the other
Nerve hook accesses "tough to reach" corners of the knee
Particularly useful for use with an ultra-congruent polyethylene insert, where trial liners
are typically not used, once the final components have been placed
Ultra hard titanium nitride coating helps to extend life by increasing surface hardness,prolonging sharpness, and resisting chemicals and corrosion

PRODUCT NO:

5007

Overall Length: 9.25" (23,5 cm)
Blade Width at End: 5 mm
Hook Depth: 5 mm

Long Bonney Tissue Forceps

Extra length—3" (7,6 cm) more thanstandard—allows for use in deep wound areas

PRODUCT NO:

5040

Overall Length: 10" (25,4 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Charnley Type Tissue Needle Forceps
Designed by Amal Das Jr., MD

Helpful for wound closure in deep
areas with fascia under tension suchas hip or knee replacement
Can also help retrieve a needle in a tight area.
PRODUCT NO:

1165

Overall Length: 6.875" (17,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Adson Forceps with Cobb Elevator End

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Designed by Oscar Castro-Aragon, MD

Has the advantages of having a Cobb
tip at the end of an Adson forceps

PRODUCT NO:

1166

Overall Length: 4.75" (12,1 cm)
Tip Width: 2.4 mm (2,4 mm)

Allows the opportunity to do soft tissue dissection, cleaning of thebone or bone fragments in a fracture, push bone fragments to holda reduction in a fracture, separate soft tissue, and turn it around topick up tissue without having to switch instruments back and forth.

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

31

Wilke Angled
Blunt Nose Scissors
Designed by Benjamin K. Wilke, MD

Allows blunt dissecting
around critical structures(nerves, vessels, etc.)
while maintaining a cutting
surface for fascia. The
tool's blunt ends can also
be used for cauterizing andgrabbing small vessels.

PRODUCT NO:

MADE EXCLUSIVELYFOR INNOMED IN

3078

Overall Length: 6" (15,2 cm)

GERMANY

Orthopedic Needle
Holder/Scissors
Drive a needle andcut a suture withoutchanging instruments
5.5" Tungsten
Carbide #3055

6.5" Tungsten
Carbide #3065

7" Tungsten
Carbide #3075

7" Standard#3070

Scissors
Holding
Tips
MADE FOR INNOMED IN

GERMANYPRODUCT NO’S:
Standard Tips

3070 7.0" (17,8 cm)
Tungsten Carbide Tips

3055 5.5" (14 cm)
3065 6.5" (16,5 cm)
3075 7.0" (17,8 cm)

Longer sizes arehelpful in orthopedics

Rogozinski Locking Needle Driver/Scissors

Designed with a quick lock & release handle, can drivea needle and cut a suture without changing instruments
Standard #3083

Scissors
Holding Tips

Large #3084
PRODUCT NO’S:

3083 [Standard]

Designed by
Chaim Rogozinski, MD

3084 [Large]

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 6.5" (16,5 cm)

Overall Length: 7.75" (19,7 cm)

GERMANY

Sweed Dissecting Scissors
Designed by Tamer Sweed, FRCS (Orth)

Designed with a blunt, flat bar
fixed to the lower limb, the scissors
also act as a dissector to protect
underlying vital structures
PRODUCT NO:

3081

Overall Length: 6.625" (16,8 cm)
Bottom Pad: 16 mm x 6 mm)
Pad Extension Beyond Scissor: 6 mm

32

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYINNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Bates Needle Holder with Suture Cutter

No switching between needle
driver and scissors, or need forassistant to cut sutures for you
Cutting with opening of forceps
reduces possibility of damage
to surrounding tissues
Sliding the instrument down to
the suture knot allows quick andconsistent 2 mm suture tails
Slip the suture strands into
the suture cutting slot andslide the closed instrument
along until desired length oftail is achieved, then open the
instrument to cut the sutures

Designed by James E. Bates, MD

By trapping the suture and cutting when theforcep is opened, helps to reduce stress on
the surgeon’s hand

PRODUCT NO:

3071

Overall Length: 8.125" (20,6 cm)
Jaw Width: .25" (6,4 mm)
Open Jaw Length: .5" (12,8 mm)

Suture Cutter
Needle Holder

Stanton Needle Driver
Designed by John L. Stanton, MD, FACS

Allows a heavy cutting needle such as anOS-6 to be pushed through cancellous
bone when re-attaching muscle or tendon
The groove captures the outer (convex) side of
the needle and prevents the needle from spinningeven when applying significant pressure. Useful forreattaching the rotator cuff in rotator cuff repairs,
as well as in attaching suture anchors.
PRODUCT NO:

3042

Overall Length: 6.75 (17,1 cm)
Jaw Width: .25" (6,3 mm)

PRODUCT NO’S:

4269-00 [Set – 1 Gelpi & 1 Blade]Also available Individually:

4269-01 [Gelpi]

Overall Length: 7.5" (19,1 cm)
Maximum Spread Width: 3.625" (9,2 cm)

4269-02 [Blade]

Overall Length: 5.5" (14 cm)
Blade Width: 1" (2,54 cm)
Blade Depth from Bend: 1.5" (3,8 cm)
Blade Bend-Back Angle: 130°

Stulberg
Incision Close
Gelpi & Blade Set

Designed by S. David Stulberg, MD

Designed to help expose
difficult to visualize areasat the end of incisions

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

33

Incision Aligner

Designed by DMP

Designed to align an incision during closing
PRODUCT NO:

1330

Overall Length: 14" (35,6 cm)
Blade Offset: 45 mm

The bent ends of the aligner areplaced at each end of an incision,
which is aligned by pulling outward
on each end. The sliding end willlock in place when it is tensioned.
Pressing inward slightly on thesliding end will allow the aligner tobe collapsed and removed.

Dodson Extremity Skin Saver
Designed by Mark A. Dodson, MD

Designed to help protect the patient's skinwhen removing a disposable tourniquet

PRODUCT NO:

8628

Overall Length: 4.75" (12,1 cm)
Width: 1.5" (3,8 cm)
Lip: .5" (1,3 cm)

Desai Surgical Funnel
Designed by Sarang Desai, DO

Helps with control and placementof bone graft or antibiotic beads
Made from surgical grade stainless steel (for sterilization).
PRODUCT NO:

8989

Profile View

Overall Length: 6.25" (15,9 cm)
Handle Length: 3.25" (8,3 cm)
Funnel Diameter at Top: 3" (7,6 cm)
Funnel Flow-thru Diameter: 11 mm

Surgical Spoon
Designed by David Scott, MD

Very useful for the application of
methylmethacrylate bone cement
Made from surgical grade stainless steel(for sterilization purposes).
PRODUCT NO:

8209

Overall Length: 5.875" (14,9 cm)

34

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Soft Tissue
Protector

Three trephine tip sizes
Trephine
DilatorK-wireWire
Repositioning
Guide Tip

Cheng Biopsy Trephine System
Designed by Edward Cheng, MD

Cannulated T-handle and trephines
allow use of a standard 1.6 mm (.062")
threaded K-wire to help facilitate grasping
and removal of a core bone sample forbiopsy or core decompression
Allows use of trephine at oblique angles to bone surface byusing an anchoring K-wire and cannulated trephine
Avoids “skipping” of trephine teeth on bone surface
Facilitates optimal approach angle and direction of trephine
Variety of core diameters yields bone samples of sufficient sizefor pathology
Adapters allow for use of a power drill
Minimally invasive — soft tissue sleeve protects surrounding
structures and tissue
Can also be used for bone graft harvesting
Repositioning guide allows easy adjustment of targeting K-wireTrephines(3 Sizes)

Dilators(4 Sizes)

Wire
Repositioning
Guide

PRODUCT NO’S:

1425-00 [Complete Set with Case]
Set Includes/Available Separately:

DriverRetraction
HandleTrephine
Drivers1425-01 [Soft Tissue Protector – Small]
1425-02 [Soft Tissue Protector – Medium]
1425-03 [Soft Tissue Protector – Large]
1425-04 [Dilator – 4.75 mm]
1425-05 [Dilator – 6.25 mm]
1425-06 [Dilator – 7.75 mm]
1425-07 [Dilator – 9.25 mm]
1425-08 [Trephine – Small]Soft
TissueProtectors(3 Sizes)

Internal Diameter: 5mm
Overall Length: 7.125" (18,1 cm)(3 Sizes)

1425-09 [Trephine – Medium]
Internal Diameter: 6.5 mm
Overall Length: 7.125" (18,1 cm)

1425-10 [Trephine – Lage]
Internal Diameter: 8 mm
Overall Length: 7.125" (18,1 cm)

1425-11 [Drive End – Small]
1425-12 [Drive End – Medium]
1425-13 [Drive End – Large]
1425-14 [Driver Retraction Handle]

Includes (2) Handle Retaining Screws (#1425-14-B-COMP)

1425-15 [3-Hole Wire Repositioning Guide]1425-Case [Case]
Replacement Part:

1425-14-B-COMP [Handle Retaining Screw]Complete Set in Case

1.800.548.2362       JANUARY 2024   

K-wire not included.

   KNEE INSTRUMENTS

35

Soft Impact Mallets

PRODUCT NO’S:

7820 [2 lbs. Standard]with Easy Grip Handles

2 lbs. Standard

Weidman handle designed by Kevin Weidman, MD#7820

Provides shock-absorbing force

Weight: 2 lbs. (.907 kg)
Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7821 [2 lbs. w/Weidman Handle]

Filled with a shock-absorbing media andhas a flat striking surface to keep the mallet
centered on an instrument while providing
less bounce or wasted force.

2 lbs. w/Weidman Handle#7821

The comfortable Easy Grip handle is madeof a textured silicone that helps prevent thesurgeon’s gloved hand from slipping andhelps maintain a solid grip. The bottom canalso be used to tap an implant in place.

2 lbs. w/Delrin End#7832

The mallet with delrin head features a
replaceable delrin head.

Weight: 2 lbs. (.907 kg)
Overall Length: 10.625" (27 cm)
Grip Length: 5.5" (14 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7832 [2 lbs. With Delrin End]
Weight: 2 lbs. (.907 kg)
Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.375" (3,5 cm)

7837 [3 lbs. Standard]

Weight: 3 lbs. (1.35 kg)
Overall Length: 11" (27,9 cm)
Handle Length: 5" (12,7 cm)
Head Width: 3.5" (8,9 cm)
Head Diameter: 1.875" (4,8 cm)

Delrin Head Replacements for 7832:

3 lbs. Standard#7837

Replacement Delrin Heads

Easy Grip

Comfortable grip helps
prevent the surgeon’s
gloved hand fromslipping and helps
maintain a solid grip.

TexturedSoft Silicone

Handles

Ortho Mallets

Small #7810

PRODUCT NO’S:

7810 [Small]

Overall Length: 8" (20,3 cm)
Handle Length: 4.5" (11,4 cm)
Head Weight: 1 lb. (.45 kg)
Head Diameter: 1.3125"

with Easy Grip Handles
These solid stainless steel mallets
each have a comfortable 4½" grip
made of a textured silicone thathelps prevent the surgeon’s gloved
hand from slipping and helps
maintain a solid grip.

7832-HEAD01 [.5" Stud] Single
7832-HEAD02 [.5" Stud] 3-Pack
7832-HEAD03 [.875" Stud] Single
7832-HEAD04 [.875" Stud] 3-Pack7815 [Large]

Overall Length: 8" (20,3 cm)
Handle Length: 4.5" (11,4 cm)
Head Weight: 1.75 lb. (.8 kg)
Head Diameter: 1.5" (3,8 cm)

Large #7815

Jones Mallet

PRODUCT NO:

Designed by Dickie Jones, MD

7825 [2.4 lbs]

Unique hand fitting shape provides
superior gripping strength

Overall Length: 8.25" (21 cm)
Head Width: 3" (7,6 cm)
Head Diameter: 1.5" (3,8 cm)

This striking instrument has a unique hand
fitting shape that provides superior gripping
strength for accurate light to heavy impaction.

36

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Bechtold Ergonomic
Orthopedic Mallet

Stainless steel head and shaft with an aluminum handle
with a right-handed grip
Large and small striking heads with smooth surface
Palmar side of the mallet features a flat surface to slide
along a broach or impacting type instrument for backslapping and serves well as an additional striking surface

Designed by Dustin Bechtold, MD

Ergonomically designed for forward
and backward strikes, featuring anergonomic handle with a tamp
PRODUCT NO:

7822 [2.7 lbs]

Weight: 2.7 lbs (1.22 kg)
Overall Length: 10.75" (27,3 cm)
Head Width: 4" (10,2 cm)
Large Head Diameter: 2" (5,1 cm)
Small Head Diameter: 1.5" (3,8 cm)

Straight Suture Passer
Designed by Brian T. Maurer, MD

Designed to help pass suture through bone
PRODUCT NO:

1111

Overall Length: 8.125" (20,6 cm)
Handle Length: 4.25" (10,8 cm)
Shaft Diameter: 2,5 mm

Ortho Suction Tube
Designed by T. Eickmann, MD

Very effective for suction and minor retracting
Helps eliminate plugging due to bone, cement fragments, blood clots, etc.
PRODUCT NO:

5465

Overall Length: 9.25" (23,5 cm)
End Hole Dia.: 1 mm
Side Hole Dia.: 1.5 mm

Beicker Curette Suction Device
Designed by Clint Beicker, MD

Designed to help visualization of afracture site within a fracture hematoma
Also useful for arthroscopic curettage of osteochondral lesions.
PRODUCT NO:

4231

Overall Length: 10.5" (26,7 cm)
Curette Cup: 7.5 mm x 5.5 mm

1.800.548.2362       JANUARY 2024      KNEE INSTRUMENTS

37

Large T-Handle
Fixed Drivers

Large easy grip soft silicone
handled drivers help provide
a sturdy non-slip grip
The two standard Quick-connect
models release by pulling thecollar backward, while the Reverse
Quick-connect model is designed
to have the collar be pushedforward for release.

Zimmer Hall
Quick-connect

HudsonQuick-connect

PRODUCT NO’S:

8248 [Zimmer Hall Quick-connect]
Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

Zimmer Hall
Quick-connect#8248

8248-01 [Reverse Quick-connect Zimmer Hall]
Overall Length: 5.75" (15,6 cm)
Handle Width: 4.625" (11,6 cm)

Zimmer Hall
Reverse Quick-connect8249 [Hudson Quick-connect]

HudsonQuick-connect#8248-01

Overall Length: 6.75" (17,1 cm)
Overall Length with Pin In Handle: 11.5" (29,2 cm)#8249

T-Handle Chuck & Key
PRODUCT NO’S:

8247-00 [T-Handle Chuck & Key]8247-01 [T-Handle Chuck Only]
8247-02 [Chuck Key Only]

Gray Syringe Assist with Ergonomic Handle

Designed by Robert Gray, MD

For use in the O.R or the office, the design helps to prevent hand fatigue
and pain when injecting with a 20mL syringe over multiple cases
Sterilizable for O.R use, such as injecting the posterior capsule during TKA
Especially useful for injecting preoperative local anesthesia for WALANT surgery
Uses finger flexors to generate more force over more surface area than only the thumb flexor
Ratchet mechanism ensures maximal grip force generation throughout entire injection

Syringe not included.

PRODUCT NO:

8988

Overall Length – Closed: 5.25" (13,3 cm)
Overall Length – Open: 7.5" (19,1 cm)
Height: 5" (12,7 cm)
Syring Diameter: 21 mm

Patent Pending

White Aspiration Handle

Designed by Edward White, MD

Designed for aspiration of cavities or spacesthat have greater than 20 ml volume, suchas joints, bone marrow, and the illiac crest
Works with a 60 ml syringe (not included) only.
Syringe not included.

PRODUCT NO:

1131

Height: 5" (12,7 cm)
Length: 6.5" (16,5 cm) / Extends to 11" (27,9 cm)
Width at Syring Holder: 1.5" (3,8 cm)
Body Width: .9" (2,3 cm)

38

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Universal Screwdriver Set
Set consists of one handle
and one sterilization/storage
case, plus seven double
ended screwdriver bits:

Helps eliminate the openingof multiple sterile packs whena specific size or style ofscrewdriver is needed

Handle#5195-01

Helpful during revision total joint surgery wherescrews have been used, removal of bone plates,
fracture fixation screws or bone graft screws.
PRODUCT NO’S:

Small & Large Single Slot #5195-02

5195 [Complete Set with Case]

Cross & Cruciate #5195-03

5195-01 [Handle]
5195-02 [Straight (single slot)]

Also sold individually

Large: 7 x 1.5 mm, Small: 5 x 1 mm

3.5 mm & 4.5 mm Hex #5195-04

5195-03 [Cross/Cruciate]

Small & Large Phillips #5195-05

5195-04 [Hex]Large: 7 mm, Small: 6 mm

Large: 4.5 mm, Small: 3.5 mm

5195-05 [Phillips]#6 & #8 star #5195-08

Large: 4 mm, Small: 3.5 mm

5195-08 [Small Star: #6 & #8]
5195-06 [Medium Star: #10 & #15]
5195-07 [Large Star: #20 & #25]

#10 & #15 star #5195-06#20 & #25 star #5195-07

Star Bit Driver Set
Helps eliminate theopening of multiple
sterile packs when aspecific size of star
bit is needed

T25

PRODUCT NO’S:

5194-00 [4 Star Bits w/Handle & Case]
5194-01 [4 Star Bits w/Case only]

T20

Set in Storage Case

#5194-25#5194-20

Also sold individually:

T15

S0113 [Universal 4" (10,2 cm) Handle]
5194-10 [T10 with A/O End]5194-15 [T15 with A/O End]5194-20 [T20 with A/O End]5194-25 [T25 with A/O End]
9003 [Case]#5194-15

T10#5194-10

Torx/Hex Adapter Set

PRODUCT NO’S:

8003-00 [Set – One Each]

Designed by Stephen M. Walsh, MD

Set Includes/Available Separately:

Designed for conversion
of a 3.5 mm screwdriver

8003-01 [Torx Bit to Hex Driver Adapter]
Overall Length: .6" (1,54 cm)

8003-02 [Hex Bit to Torx Driver Adapter]
Overall Length: .6" (1,54 cm)

Helpful during revision total joint surgery. Set
consists of four star bits — T10, T15, T20, &T25, a handle which accommodates any of
the above bits, and a sterilization case. The
drive end (A/O) is designed for easy and quick
engagement with the universal instrument
handle. The ergonomic, modular handle has twoconnection points, allowing for both straightand T-handle orientations.

Especially helpful when an articulated, universal
joint driver is needed (i.e. acetabular screws)

Hex Bit to Torx Driver Adapter#8003-02

Torx Bit to Hex Driver Adapter#8003-01

Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

1.800.548.2362       JANUARY 2024   

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:   KNEE INSTRUMENTS

39

PRSRT. STD.
U.S. POSTAGE
PAID
GRAND RAPIDS, MI

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 748

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

TKA Gap Assessment Gauge Assembly
Designed by Michael Radon

Universal design allows the gauge to be used without
the removal of trials to help determine if a 1 or 2 mm
additional thickness insert may be needed
Alignment rod can be inserted in thegauge to help check alignment.

PRODUCT NO’S:

5216-00 [Assembly]

Gauge

Individual/Replacement Parts:

5216-01 [Gauge]

Overall Length: 7.5” (19,1 cm)
Width: 2.5” (5,4 cm)
Prong Length: 2” (5,1 cm)
Prong Width: (.75” (1,9 cm)
Gap Between Prongs: 1” (2,5 cm)

Alignment Rod

5216-02 [Alignment Rod]Overall Length: 18” (45,7 cm)
Diameter: .1875” (4,75 mm)

INNOMED, INC

103 Estus Drive
Savannah, GA 31404

ISO 13485:2016

Innomed-Europe LLC

Innomed-Europe GmbHTel 912.236.0000
Fax 912.236.7766

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74

innomed.net
info@innomed.net

www.innomed-europe.com
orders@innomed-europe.com

www.innomed-europe.com
orders@innomed-europe.com

Villingen-Schwenningen,Deutschland
Tel 0049 (0) 7720 46110 60

1.800.548.2362   INNOMED.NET© 2024 Innomed, Inc.
All Rights Reserved

0124