J AANPURAI LR Y

2 0 2 43

OR T HOPEDIC INS T RUMEN T S®

Ortho Self-Retaining
Retractors

PCL Retractor – Straight
Page 3

Page 19

Chandran Modified
Knee Retractor

Featuring
many

instruments
throughout

Page 11“S” Total Knee
Retractors
Page 8

Powers Flared
Small Knee Retractor
Page 5Chandran Thigh Lift Positioner
Page 26

Knee Exposure & Positioning
1.800.548.2362   INNOMED.NET

Knee Retractor System
Designed by S. David Stulberg, MD

Helps free assisting
personnel while providing
excellent exposure

Wide PCL Retractor with Strap
PRODUCT NO:

3525

Overall Length: 10" (25,4 cm)
Blade Width Above Prongs: 57 mm
Prong Width: 8,5 mm | 17 mm Gap | 8,5 mm

MIS Modified Wide PCL Retractor with Strap
PRODUCT NO:

3515

The Knee Retractor System holds retractors utilizing Velcro
straps. This helps eliminate obstruction of the surgeon’s
operative area and frees assisting personnel. Four retractor
styles are available; straps are available in two lengths.
Retractors and straps are autoclavable. The Retractors
can be used singularly or in combination.

®

Overall Length: 10" (25,4 cm)
Blade Width Above Prongs: 34 mm
Prong Width: 8,5 mm | 17 mm Gap | 8,5 mm

Strap
PRODUCT NO’S:
Packages of 10

8100-P [Long Strap–Femur]
8120-P [Short Strap–Tibia]

Velcro® is a registered trademark of Velcro U.S.A.

Single Prong Collateral Ligament Retractor with Strap
PRODUCT NO:

6650

Overall Length: 8.25" (21 cm)
Blade Width: 14 mm

Long Prong Collateral Ligament Retractor with Strap
PRODUCT NO:

6630

Overall Length: 8" (20,3 cm)
Overall Blade Width: 21 mm
Prong Width: 4,5 mm | 12 mm Gap | 4,5 mm

Stubbs Short Prong Collateral Ligament Retractor with Strap
PRODUCT NO:

6640

Designed by B.
Stubbs, MD

Overall Length: 8" (20,3 cm)
Blade Width Above Prongs: 27 mm
Prong Width: 4,8 mm | 3,4 mm Gap | 4,8 mm

MIS PCL Retractor
Designed by S. David Stulberg, MD

2

PRODUCT NO:

6203 [Medium]

Overall Length: 12.5" (31,8 cm)
Handle Length: 6" (15,2 cm)
Blade Width: 15 mm

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

PCL Retractors

Designed to straddle
the cruciate ligamentand lie in the condylar
notch, allowing thesurgeon to retract
the tibia away fromthe femur for better
accessStraight#2820-S

Standard#2820

CoatedStandard#2820-CPRODUCT NO’S:

2820-S [Straight]

Overall Length: 10.375" (26,3 cm)
Prong Width: 5 mm | 10 mm Gap | 5 mm

2820 [Standard]

OrthoLucent™
Standard#2820-R

Overall Length: 9.875" (25,1 cm)
Prong Width: 5 mm | 10 mm Gap | 5 mm

2820-C [Coated Standard]

Overall Length: 9.875" (25,1 cm)
Prong Width: 5 mm | 10 mm Gap | 5 mm

2820-R* [OrthoLucent™ Standard]

Wide Prong#2825

The handle is contoured away from the surgeon’s field of view.
Wider prongs offer better stability
Modular weights can be used to help hold the retractor in place.
Mayo Wide Prong
The Coated Standard PCL includes a special protective coating, applied towith Ergonomic Handle
the areas of the instrument that may come into contact with component surfaces,#2825-01
to help prevent from marring the articulating surfaces.
The OrthoLucent™ Standard PCL can be safely used to look behind the knee when the component(s) are inplace without metal transfer or marring component surfaces when contact is made. It is made of a strong,lightweight carbon fiber PEEK composite material, which is completely radiolucent, and can be steam sterilized.

MIS Modified Wide PCL Retractor
Designed by S. David Stulberg, MD

Overall Length: 9.875" (25,1 cm)
Prong Width: 5 mm | 10 mm Gap | 5 mm

2825 [Wide Prong]

Overall Length: 9.875" (25,1 cm)
Prong Width: 8,5 mm | 11 mm Gap | 8,5 mm

2825-01 [Mayo Wide Prong with
Ergonomic Handle]

Overall Length: 11" (27,9 cm)
Prong Width: 8,5 mm | 11 mm Gap | 8,5 mm∞ Designed by Joseph Mayo, MD

Handle designed by Munish C. Gupta, MD
EXCLUSIVELY* MADEFOR INNOMED IN

SWITZERLAND

Modular Weights

Weights can be used to help
hold the retractors in place.
PRODUCT NO’S:

3430-01 1.5 lbs. (.68 kg)
3430-02 2.0 lbs. (.91 kg)
3430-03 2.5 lbs. (1.13 kg)

PRODUCT NO’S:

3510 [Standard]

Overall Length: 10" (25,4 cm)
Blade Width Above Prongs: 34 mm
Prong Width: 8,5 mm | 17 mm Gap | 8,5 mm

with attaching hook3515 [With Velcro Strap]

Overall Length: 10" (25,4 cm)
Blade Width Above Prongs: 34 mm
Prong Width: 8,5 mm | 17 mm Gap | 8,5 mm

Wide PCL Retractor
Designed by S. David Stulberg, MD

Helps expose the proximal
tibia for better surface access
Designed to expose the proximal tibia during
total knee surgery for better access to thearticulating surfaces. The handle is contoured
to allow the surgeon a clear field of view of theoperating area. Modular weights can be used
to help hold the retractor in place.

1.800.548.2362       APRIL 2024   

PRODUCT NO:

3520

Overall Length: 10" (25,4 cm)
Blade Width Above Prongs: 57 mm
Prong Width: 8,5 mm | 17 mm Gap | 8,5 mm    KNEE EXPOSURE & POSITIONING INSTRUMENTS

3

Rosen Knee Tibial Retractor

PRODUCT NO:

2830

Overall Length: 12.5" (31,8 cm)

Designed modification by Adam Rosen, DO of original design by Christopher M. Meckel, MD

Designed for total knee and revision total knee replacements
using posterior stabilized knee components, the posterior build-up
on the retractor allows the surgeon to more easily translate the tibia
forward for better visualization after femoral notch preparation

Meckel Posterior
Stabilizing Knee Retractor
Designed by Christopher M. Meckel, MD

Designed to provide enhanced anterior translation of thetibia when doing posterior stabilized total knee replacement
The 15 mm deep blade section of the retractor is used to lever the tibia forward (by resting
the tip on the posterior tibia and the middle blade section block levering off the distal
femur) after the box cut has been made in the distal femur.
PRODUCT NO:

4538

Blade Width: 20 mm
Blade Depth: 15 mm
Overall Length: 10" (24,5 cm)

Manzary Proximal Tibial
Stabilizing Knee Retractor
Design modification by Mojieb Manzary, MD, FRCS of original
design by D. Kevin Lester, MD, and Christopher M. Meckel, MD

Designed to help subluxing the tibia anteriorly inposterior stabilizing total knee replacement, helping
to expose the proximal surface for preparation
PRODUCT NO:

4531

Overall Length: 12.75" (32,4 cm)
Blade Width: 1.5" (3,8 cm)
Block Dimensions: .8" x 1.5" x .5"       (2 x 3.8 x 1.25 cm)

PRODUCT NO:

4699

Overall Length: 12" (30,5 cm)
Depth from Bend: 5" (12,7 cm)
Blade Width: 1.5" (3,8 cm)

4

Helps expose the cut
surface of the tibia to allow
sizing, preparation andcleansing during TKA

INNOMED.NET   Lester Proximal
Tibial TKA Retractor
Designed by D. Kevin Lester, MD

Also helps protect the posterior knee
soft tissue structures from injury.

   FREE TRIAL ON MOST INSTRUMENTS

Minimally Invasive Knee Retractors
Helps provide excellent visibility andligament protection during Total andUnicondylar Knee Replacement Surgery

PRODUCT NO’S:

1
1

2

3

4

5

4

Minimally Invasive
Small Hohmann Retractor

2

3

Minimally Invasive
Condylar Retractor

4#S3037Overall Length: 8" (20,3 cm)
Blade Width: 36 mm

S3037 [Condylar Retractor]
Overall Length: 7.5" (19,1 cm)
Blade Width: 12 mm

S3038 [Superior Retractor]
Overall Length: 8.75" (22,2 cm)
Blade Width: 31 mm

S3039 [Patellar Retractor]

Minimally Invasive
Large Hohmann Retractor

5#S3036#S3035Overall Length: 7.5" (19,1 cm)
Blade Width: 25 mm

S3036 [Large Hohmann Retractor]
2
3

1

S3035 [Small Hohmann Retractor]

Minimally Invasive
Superior Retractor

Overall Length: 10.25" (26 cm)
Blade Width: 45 mm

5Minimally Invasive
Patellar Retractor#S3039#S3038

Powers Flared Small Knee Retractor
Designed by Mark Powers, MD

A bent knee retractor with a cobra flareto help provide optimal exposure
PRODUCT NO:

6291

Overall Length: 8" (20,3 cm)
Pad Flare Width: 17 mm
Point Length Below Pad: 21 mm

45° Knee Retractors

Small#6290-00-076

Designed for use around the knee
PRODUCT NO’S:

Medium

6290-00-075 [Large]#6290-00-077

Overall Length: 9.125" (23,2 cm)

6290-00-076 [Small]

Overall Length: 7.875" (20 cm)

6290-00-077 [Medium]

Overall Length: 9.125" (23,2 cm)

Medium Straight

6290-00-078 [Medium Straight]#6290-00-078

Overall Length: 9.125" (23,2 cm)

Large#6290-00-075

Chandran Tibial Knee Retractor
Designed by Rama E. Chandran, MD

Designed for use in TKR, the hook on
the front of the blade acts as a stop to
help prevent the retractor from deep
penetration behind the tibia
PRODUCT NO:

4533

Overall Length: 10.5" (26,7 cm)
Depth from Bend: 3" (7,6 cm)

1.800.548.2362       APRIL 2024       KNEE EXPOSURE & POSITIONING INSTRUMENTS

5

Harwin Modified Cobra Retractor

Designed by Steven F. Harwin, MD, FACS

Designed for use during total knee and hip surgery
The long handle and obtuse angle provide more ergonomic leverage—especially helpful
for use with obese patients.
For total knee surgery, the wide blade of the large retractor spans the preparedbox and helps bring the tibia forward. The small retractor helps with retraction of themedial and lateral structures, where the wide, concave blade provides added exposure
over standard bent Hohmann retractors. The serrated tip helps improve stability.
In total hip surgery, the wide, concave blade design allows for enhanced
exposure—especially useful in anterior hip surgery with the placement of reamers, andto elevate and expose the proximal femur.
PRODUCT NO’S:

6143 [Large]

Overall Length: 14.75" (37,5 cm)
Horizontal Handle: 8" (20,3 cm)
Blade Width: 43 mm
Tongue: 25 mm x 5 mm

6143-01 [Small]

Overall Length: 12.5" (31,8 cm)
Horizontal Handle: 5" (12,7 cm)
Blade Width: 30 mm
Tongue: 25 mm x 5 mm

Small

Large

#6143-01#6143

Narrow Right Angle Retractor
Designed for soft tissue retraction
PRODUCT NO:

C1011

Overall Length: 8.5" (21,6 cm)
Handle Length: 6.75" (17,1 cm)
Blade Depth: 4.5" (11,4 cm)
Blade Width: .375" (1 cm)

Taylor Retractors

Modular Weights

PRODUCT NO’S:

Weights can be used to help
hold the retractors in place.

6330-01 [Standard]

Overall Length: 8" (20,3 cm)
Depth from Bend: 4" (10,2 cm)
Blade Width: 32 mm

PRODUCT NO’S:

3430-01 1.5 lbs. (.68 kg)
3430-02 2.0 lbs. (.91 kg)
3430-03 2.5 lbs. (1.13 kg)

6330-02 [Large]

Overall Length: 9" (23 cm)
Depth from Bend: 5.5" (14 cm)
Blade Width: 32 mm

6330-03 [Deep with Pin Guides]
Overall Length: 9" (23 cm)
Depth from Bend: 5.5" (14 cm)
Blade Width: 32 mm
Guide for Pins Up To: .125" (3,2 mm)with attaching hookStandard#6330-01

Deep with Pin Guides#6330-03

Deep#6330-02

McPherson Retractor Extender

Designed by Ed McPherson, MD

Designed to extend a standard retractor
to help provide additional leverage
PRODUCT NO'S:
Available in two sizes to accommodate most retractors— standard for retractors up to .125" (3,2 mm) thick,and large for retractors up to .16" (4 mm) thick.

66022 [Standard]

Overall Length: 15.625" (39,7 cm)
For retractors up to .125” (3,2 mm) thick6022-01 [Large]

Overall Length: 15.625" (39,7 cm)
For retractors up to .16” (4 mm) thick

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

BoothKnee Retractor
Designed by Robert E. Booth, Jr., MD

Designed to help protect the tibial
surface and to tighten the collateral ligaments
and to help assess the rotation of the femur
PRODUCT NO:

6580

Overall Length: 11.75" (29,8 cm)
Handle Length: 10.5" (26,7 cm)
Blade Depth: 2.75" (7 cm)
Blade Width: 2" (5,1 cm)

Glassman
Posterior Condylar
Osteophyte Retractor
Designed by Andrew Glassman, MD

Designed to provide exposureof the posterior condyle to gain accessto posterior condylar osteophytes during
unicompartmental and total knee arthroplasty

PRODUCT NO:

3730

Overall Length: 11.75" (29,8 cm)
Depth From 90° Bend: 3" (7,6 cm)
Length from 45° Bend: 1" (2,54 cm)
Blade Width: 1" (25,4 mm)

Chandler Retractor

Used for retracting tissue away from the bone, andhelpful for posterior exposure of the tibia in MIS surgery

PRODUCT NO’S:

1,6 cm #3220-01
1,9 cm #3220-02
2,54 cm #3220-03

3220-01 [5/8"]

Overall Length: 9.875" (25,1 cm)
Blade Width: 5/8" (1,6 cm)

3220-02 [3/4"]

Overall Length: 9.875" (25,1 cm)
Blade Width: 3/4" (1,9 cm)

3220-04 [1"]Overall Length: 9.875" (25,1 cm)
Blade Width: 1" (2,54 cm)

3220-02R* [OrthoLucent™ 3/4"]
Overall Length: 9.875" (25,1 cm)
Blade Width: 3/4" (1,9 cm)

Allows the surgeon to retract soft tissue away from the bone, and canbe used for knee, hip and shoulder surgery. The specially designed handle
is contoured away from the surgeon’s field of view and working area.
The OrthoLucent™ version is made of a lightweight carbon fiber PEEK composite material,
which is strong, lightweight, completely radiolucent, helps to prevent from marring
component surfaces, and can be steam sterilized.

OrthoLucent™ 3/4"

#3220-02R

EXCLUSIVELY* MADEFOR INNOMED IN

SWITZERLAND

Bolanos Modified
Chandler Retractor
Designed by Alberto Bolanos, MD

Used for retracting tissue
away from the bone

1.800.548.2362       APRIL 2024   

PRODUCT NO:

3222

Overall Length: 7.5" (19,1 cm)
Blade Width at Widest: 1" (2,54 cm)    KNEE EXPOSURE & POSITIONING INSTRUMENTS

7

“S” Total Knee Retractors
Designed by R. Barry Sorrells, MD

Helps protect the collateral ligaments
and popliteal structures while providing
excellent visualization within the knee joint

Wide#3720-00

The design is self-retaining and can be used singularly and inpairs. For cruciate sparing or sacrificing prosthetic designs.
PRODUCT NO’S:

3720-00 [Wide Blade]

Overall Length: 6" (15,2 cm)
Blade Width: 20 mm

Excellent for minimallyinvasive knee procedures

3720-01 [Narrow Blade]
Overall Length: 6" (15,2 cm)
Blade Width: 10 mm

Narrow#3720-01

Rosen Double Ended
Army-Navy/"Z" Retractor
Designed By Adam Rosen, DO

Helps to reduce the number of
instruments on the field andto limit the need for passing
instruments during the case

PRODUCT NO:

4005

Overall Length: 10" (25,4 cm)
Z End: 70 mm Deep, 11 mm Wide
Army Navy End: 40 mm Deep, 15 mm Wide

Rosen Double Ended
Richardson Retractor
Designed by Adam Rosen, DO

Designed to help with exposureand soft tissue protection

PRODUCT NO:

4010-01

Overall Length: 10" (25,4 cm)
Large End: 41 mm x 40 mm
Small End: 38 mm x 38 mm

Rosen Double Ended
Richardson/"Z" Retractor
Designed By Adam Rosen, DO

Can be used in a variety of positions
for exposure, soft tissue protection,
and when making cuts

PRODUCT NO:

4010

Overall Length: 10" (25,4 cm)
Z End: 70 mm Deep, 11 mm WideRichardson End: 40 mm Deep, 37 mm Wide

Uni Medial/Lateral Ligament Retractor
Designed by Kurt Kramer, PA-C

Designed to be placed in the medial/lateral tibial
recess while making the horizontal tibial cut during
unicompartmental knee arthroplasty—helping to retract
and protect the medial and lateral collateral ligaments
Ambidextrous, ergonomic design allows for comfortable and natural
hand positioning, helping to improve MCL/LCL protection and easeof use, especially in the obese patient.
PRODUCT NO:

3632

Overall Length: 4.25" (10,8 cm)
Blade Width: 8.8 mm
Blade Depth: 2.375” (6 cm)

8

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

“Z” Knee Retractor

Helps create better access to the articulating surfaces
Designed to expose the femur and the tibia during knee surgery for better
access to the articulating surfaces. The “Z” contouring of the retractor
provides the surgeon with an open field of view and working area.

PRODUCT NO:

4420-00

Overall Length: 7.25" (18,4 cm)
Blades: 11 mm Wide, 3" (7,6 cm) Deep

Engh Intercondylar Notch Retractors
Designed by Gerard A. Engh, MD

Enhances minimally invasive exposureof the medial femoral condyle
PRODUCT NO’S:

3230-01 [Small]

Blade Width at Teeth: 9 mm
Depth from Bend: 2.25" (5,7 cm)
Overall Length: 8.125" (20,6 cm)

Large#3230-03

3230-02 [Medium]

Blade Width at Teeth: 10 mm
Depth from Bend: 2.25" (5,7 cm)
Overall Length: 8.125" (20,6 cm)

Medium3230-03 [Large]

Blade Width at Teeth: 12 mm
Depth from Bend: 2.25" (5,7 cm)
Overall Length: 8.125" (20,6 cm)#3230-02

Used forunicondylar arthroplasty

Small#3230-01

Blount Knee Retractor
Designed by James B. Stiehl, MD

Designed for retraction in
total knee arthroplasty

PRODUCT NO:

The long narrow blade easily fits above the capsularligament at the joint line. Can also be used for knee
revision, fitting easily around the implant.

Blount Retractor with Small Handle
Designed by Ronald Romanelli, MD

A blount retractor with a lightweight ergo dynamic
handle designed for tissue retraction and closure
assistance in knee, shoulder, and hip arthroplasty
1.800.548.2362       APRIL 2024   

4850

Overall Length: 8.5" (21,6 cm)
Prong Width: 9 mm

PRODUCT NO:

4852

Overall Length: 9.375" (25,1 cm)
Handle Length: 4.625" (11,7 cm)
Blade Depth: 1.5" (3,8 cm)
Blade Width at Widest: .325" (8,3 mm)    KNEE EXPOSURE & POSITIONING INSTRUMENTS

9

Concave Total Knee Retractor

PRODUCT NO’S:

6720 [Standard]

Overall Length: 9.625" (24,4 cm)
Blade Width: 15 mm

6720-01 [Narrow]

Overall Length: 9.625" (24,4 cm)
Blade Width: 9 mm

Used to retract soft tissue awayfrom the femur and tibia

Used during total knee surgery to retract soft tissue
away from the femur and tibia. The blade is designed
to curve around the distal femur and tibia plateau.

Knee Retractors with
Easy Grip Handles

Retractors help provide excellent visibility
and ligament protection during total andunicondylar knee replacement surgery.

Small Hohmann Retractor

Condylar Retractor#SS3035

PRODUCT NO'S:#SS3037

Silicone handles help
reduce holding fatigue

SS3035 [Small Hohmann Retractor]
Overall Length: 7" (17,8 cm)
Blade Width: 25 mm

SS3037 [Condylar Retractor]
Overall Length: 7" (17,8 cm)
Blade Width: 12 mm

SS3038 [Superior Retractor]
Overall Length: 8.25" (21 cm)
Blade Width: 31 mm

Soft Tissue Retractor

SS3042 [Soft Tissue Retractor]

Superior Retractor#SS3042Overall Length: 8.25" (21 cm)
Blade Width: 36 mm#SS3038

MIS Patella Retractor
Designed by William Robb, MD
PRODUCT NO:

﻿3220-05

Overall Length: 9" (22,9 cm)
Patella Pad Width at Widest: 22 mm
Lower Blade Width at Widest: 16 mm

AORI Patellar Retractor

Designed by Gerard A. Engh, MD

Designed to enhance exposure during total
knee arthroplasty, the retractor has a deepbasket and two rows of teeth to grab andhold to the lateral side of the patella
The curved handle provides a fulcrum so that the applied
force will both displace and evert the patella from the femur.
The patellar retractor is placed after a routine midline,
midvastus, or medial para patellar surgical approach to theknee. Once the patella is everted the patellar retractor isapplied to the lateral border of the patella.

10

PRODUCT NO:

4690

Overall Length: 7" (17,8 cm)
Prong Width: 10 mm | 22 mm Gap | 10 mm

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Collateral
Ligament Retractor

Helps protect the lateral collateral ligament while exposing the proximal tibia
Used during total knee surgery and is inserted between thelateral collateral ligament and bone to protect the ligamentand expose the proximal tibia. The dual prongs keep theretractor from rocking and assist in the insertion. The retractor
is bent so that it is out of the way of the operating surgeon.
PRODUCT NO:

6620

Overall Length: 8" (20,3 cm)
Prong Width: 5 mm | 11 mm Gap | 5 mm

Chandran Modified Knee Retractor
Designed by Rama E. Chandran, MD

Designed to help protect the patellar tendon
during robotic assisted total knee replacement
Also useful to retract structureson the lateral side of the tibia.
PRODUCT NO:

7117

Overall Length: 8.75" (22,2 cm)
Blade Width: .625" (1,6 cm)
Blade Depth from Handle: 5" (12,7 cm)

Bicos Meniscal Repair Retractor
Designed by James Bicos, MD

A popliteal retractor specifically designed formeniscal repair or access to the posterior knee
Used when an inside out meniscal repair is indicated, the design
facilitates retracting the posterior soft tissues of the popliteal fossa
out of the way, allowing passage of meniscal repair needles.
The retractor's compact design facilitates a minimally invasive
incision. The unique shape helps capture the meniscal repair
needles and direct them out of the posterior incision for easy
grasping and repair. Incorporates a shiny body to help reflect inside
the posterior wound and aid in seeing and retrieving the needles.

PRODUCT NO:

2731

Overall Length: 5" (12,7 cm)
Depth: 1.625" (4,1 cm)
Diameter: 28 mm

Sherman Patella Tendon
Harvest Retractor
Designed by Mark Sherman, MD

Designed to help improve exposureand lessen the incision necessary toharvest a patella tendon graft during
anterior cruciate ligament bone—patella
tendon—bone (BTB) reconstruction
1.800.548.2362       APRIL 2024   

PRODUCT NO:

4691

Overall Length: 8.5" (21,6 cm)
Depth from Bend: 3.25" (8,3 cm)
Blade Width: .875" (22 mm)
Prong Length: 10 mm    KNEE EXPOSURE & POSITIONING INSTRUMENTS

11

Modified
Hohmann Retractors
Handle is contoured toallow better leverageand visualization
Useful for retracting tissues
around the bone. Can beheld in place with weights
or by hand.
Narrow#4535

PRODUCT NO’S:

4535 [Narrow]

Overall Length: 10" (25,4 cm)
Blade Width: 14 mm

OrthoLucent™ Narrow#4535-R4535-R* [OrthoLucent™ Narrow]
Overall Length: 10" (25,4 cm)
Blade Width: 18 mm

4535-01 [Extra Deep Narrow]
Overall Length: 11.625” (29,5 cm)
Blade Width: 16.4 mm

Short-tipped Narrow4545 [Short-tipped Narrow]#4545

Designed by Carl DiRaimondo, MD

Overall Length: 9.5" (24,1 cm)
Blade Width: 14 mm

6595 [Wide]

Overall Length: 10" (25,4 cm)
Blade Width: 42.5 mm

6595-01 [Extra Deep Wide]
Overall Length: 11.5" (29,2 cm)
Blade Width: 42.5 mm

The OrthoLucent™ version is made of a strong, lightweight
carbon fiber PEEK composite material, which is completely
radiolucent, helps to prevent from marring component
surfaces, and can be steam sterilized.

* MADE EXCLUSIVELY

Gain Radiolucence Without
Compromising Strength or Function
Made of a lightweight carbon fiber PEEK
composite material
Ideal for total joint use with x-ray
Steam sterilizable
Completely radiolucent
Flat black non-gloss finish
No metal transfer with component contact
Slightly higher cost than stainless steel#4535-01

Wide

#6595

Extra Deep Wide

FOR INNOMED IN

SWITZERLAND

OrthoLucent™ Retractors

Extra Deep Narrow#6595-01

PCL Retractor#2820-R

Chandler Retractor#3220-02R

Modified Narrow Hohmann#4535-R

Modified Blunt Hohmann#4550-R

PRODUCT NO’S:

2820-R [PCL]

Overall Length: 8" (20,3 cm)
Prong Width: 5 mm | 1 0mm Gap | 5 mm

3220-02R [Chandler]Standard Hohmann#4558-R

Overall Length: 9.875" (25,1 cm)
Blade Width: 3/4" (1,9 cm)

4535-R [Modified Narrow Hohmann]
Overall Length: 10" (25,4 cm)
Blade Width: 18 mm

4550-R [Modified Blunt Hohmann]

Narrow Bent Hohmann#7110-R

Blade Width at Widest: 24,5 mm
Overall Length: 10.75" (27,3 cm)

4558-R [Standard Hohmann]Blade Width: 16 mm
Overall Length: 9.625" (24,4 cm)

7110-R [Narrow Bent Hohmann]
Overall Length: 9.75" (23,8 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 19 mm
Depth from Bend: 4.75" (12,1 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

12INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Goytia Stackable Hohmann Retractors
Designed by Robin N. Goytia, MD

Interlocking design helps to increase depth andleverage in hip exposure, particularly of the anterior
acetabulum—especially useful with large patients
Standard #4551

Bent #4552

Deep Standard #4551-D

Deep Bent #4552-D

Wide #4553

Deep Wide #4553-D
DEEP VERSIONSPRODUCT NO’S:

4551 [Standard]

4552 [Bent]

4553 [Wide]

4551-D [Deep Standard]

4552-D [Deep Bent]

4553-D [Deep Wide]

Overall Length: 9.25” (23,5 cm)
Handle Length: 5.625” (14,3 cm)
Blade Width: 19.5 mm
Overall Length: 11.4” (29 cm)
Handle Length: 5.625” (14,3 cm)
Blade Width: 20 mm

Overall Length: 8.25” (20,3 cm)
Handle Length: 5.625” (14,3 cm)
Blade Width: 19.5 mm
Overall Length: 9.1” (23,1 cm)
Handle Length: 5.625” (14,3 cm)
Blade Width: 20 mm

Overall Length: 9.25” (23,5 cm)
Handle Length: 5.625” (14,3 cm)
Blade Width: 43 mm
Overall Length: 12” (30,5 cm)
Handle Length: 5.625” (14,3 cm)
Blade Width: 43 mm

2" (5 cm) deeper for use withlarge patients where extra depth,
leverage and force is needed
Custom fitted holes for interlocking retractors helps provide stability
When “stacked”, the increased lever arm of the retractor helps reduce fatigue
Ideal for use with large patients where extra depth, leverage and force is needed

Lateral Retraction Handle

for Goytia Stackable Hohmann Retractors
Design modification by Brandon Thompson, CST/CFA of original design by Robin N. Goytia, MD
PRODUCT NO:

Designed to allow lateral retraction when added
to any of the Goytia stackable hohmann retractors

Goytia stackable hohmannretractor(s) not included.

4551-HOverall Length: 9.25" (23,5 cm)

MIS Utility Knee Retractor

PRODUCT NO:

3220-03

Designed by William Robb, MD

Overall Length: 9" (22,9 cm)
Blade Width: 1,6 cm

Used interchangeably for medial
exposure, lateral exposure and to assistin posterior exposure for the tibia
Helps to keep hands out of the field of view whileproviding retraction in minimally invasive knee surgery.

Roose Utility Knee Retractor
Designed by Paul Roose, DO

Used for retraction of the soft tissues laterally
or medially and for anterior translation of thetibia during tibial prosthetic insertion
The curvature and width are designed for retraction of soft
tissues and excellent visualization of bone structure.

PRODUCT NO:

4532

Overall Length: 9" (22,9 cm)
Blade Width (above tip): 1,4 cm

1.800.548.2362       APRIL 2024       KNEE EXPOSURE & POSITIONING INSTRUMENTS

13

Teurlings Modified
Bent Hohmann Retractor
Designed by Luc Teurlings, MD

Designed to help protect the femur
cuts while retracting the MCL
The twisted blunt end also helps
elevate the femur and protect the MCL.
PRODUCT NO:

7109

Overall Length: 8.625" (21,9 cm)
Depth from Bend: 4.25" (10,8 cm)
Blade Width (Upper Portion): 17 mm
Blade Width (Lower Portion): 11 mm

Bent Hohmann
Retractors—Narrow

Helps retract tissues at themargins of the joint
The OrthoLucent™ version is made of a strong,lightweight carbon fiber PEEK composite material, whichis completely radiolucent, helps to prevent from marring
component surfaces, and can be steam sterilized.
PRODUCT NO’S:

7110 [Narrow]

Overall Length: 9.75" (23,8 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 19 mm
Depth from Bend: 4.75" (12,1 cm)

EXCLUSIVELY* MADEFOR INNOMED IN

Narrow

SWITZERLAND#7110

7110-R* [OrthoLucent Narrow]™

Overall Length: 9.75" (23,8 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 19 mm
Depth from Bend: 4.75" (12,1 cm)

OrthoLucent™
Narrow#7110-R

7110-01 [XL Handle Narrow]
Overall Length: 11.5" (29,2 cm)
Handle Length: 10" (25,4 cm)
Blade Width: 19 mm
Depth from Bend: 4.75" (12,1 cm)

Narrow with
Extra Long Handle#7110-01

7111 [Narrow w/Extra Grip Tip]
Overall Length: 9.75" (23,8 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 19 mm
Depth from Bend: 4.25" (10,8 cm)

Narrow with
Extra Grip Tip#71117115 [Short-tipped Narrow]
Overall Length: 8.625" (21,9 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 19 mm
Depth from Bend: 4.4" (11,2 cm)

Short-tipped Narrow#7115

7115-01 [Short-tipped Narrow     Extra Long Handle]
Overall Length: 11" (27,9 cm)
Handle Length: 10" (25,4 cm)
Blade Width: 19 mm
Depth from Bend: 4.25" (10,8 cm)

Short-tipped Narrow
with Extra Long Handle

7115-03 [Extra Deep Narrow]#7115-01

Overall Length: 12.125” (31,1 cm)
Handle Length: 9.75” (24,8 cm)
Depth from Bend: 6.25” (15,9 cm)
Blade Width: 19 mm

Extra Grip Tip design modification by Alfred A. Durham, MD
Short-tipped design modification by Carl DiRaimondo, MD

Extra Deep
Short-tipped Narrow

Bent Hohmann
Retractors—Wide#7115-03

Wide

PRODUCT NO’S:#6590

6590 [Wide]

Overall Length: 9.375" (23,8 cm)
Handle Length: 7" (17,8 cm)
Blade Width: 41 mm
Depth from Bend: 4.75" (12,1 cm)

Can be passed
over the marginsof the joint andheld in place withweights or by hand.

6590-01 [Extra Long Handle]
Overall Length: 11" (27,9 cm)
Handle Length: 9" (22,9 cm)
Blade Width: 41 mm
Depth from Bend: 5.5" (14 cm)

14

Wide with
Extra Long Handle#6590-01

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Modified Angled
Hohmann Retractor withLong Handle and Short Tip
Designed by R. Michael Meneghini, MD

Longer handle to help provide
safe patella retraction withexcellent ergonomics, and usefulin other orthopedic procedures
Excellent for gently retracting the patella andextensor mechanism on heavy obese patientsand muscular male patients.

PRODUCT NO:

7119

Overall Length: 11.5" (29,2 cm)
Handle Length: 9.25" (23,5 cm)
Blade Depth: 5" (12,7 cm)
Blade Width: .75" (19 mm)
Prong Depth: 1" (2,54 cm)

Hohmann Retractor

Designed like the original
Hohmann-style retractor
The OrthoLucent™ version is made of a strong,lightweight carbon fiber PEEK composite
material, which is completely radiolucent,helps to prevent from marring
component surfaces, and canbe steam sterilized.

PRODUCT NO’S:

4558 [Standard]

Blade Width: 16 mm
Overall Length: 9.46" (24 cm)

4558-R* [OrthoLucent™]*OrthoLucent™

Standard

OrthoLucent™

#4558#4558-R

Extra Deep#4558-01

MADE EXCLUSIVELYFOR INNOMED IN

Blade Width: 16 mm
Overall Length: 9.46" (24 cm)

4558-01 [Extra Deep]

Blade Width: 16,7 mm
Overall Length: 11.5” (29,2 cm)

SWITZERLAND

Modified Blunt Hohmann Retractor
Used for soft tissue retraction
The OrthoLucent™ version is made of a strong,lightweight carbon fiber PEEK composite
material, which is completely radiolucent,helps to prevent from marring
component surfaces, and canbe steam sterilized.

PRODUCT NO’S:

4550 [Standard]

Blade Width at End: 11 mm
Overall Length: 10.75" (27,3 cm)

4550-R* [OrthoLucent™]*OrthoLucent™

Standard#4550

OrthoLucent™
#4550-R

Extra Deep

#4550-01

MADE EXCLUSIVELYFOR INNOMED IN

Blade Width at Widest: 24,5 mm
Overall Length: 10.75" (27,3 cm)

4550-01 [Extra Deep]

SWITZERLAND

Blade Width at End: 11 mm
Overall Length: 13.25" (33,7 cm)

Wetzel Modified
Hohmann Retractor

Designed by Robert Wetzel, MD and Todd McKinley, MD

PRODUCT NO:

4539

Overall Length: 10" (25,4 cm)
Blade Width: .85" (21,5 mm)

1.800.548.2362       APRIL 2024   

The long point is designed to be placedaround, on, or through a bony structure
and then levered back to retract tissue
The handle is contoured to allow better leverage andvisualization. Can be held in place with weights or by hand.

    KNEE EXPOSURE & POSITIONING INSTRUMENTS

15

Baldwin LateralSoft Tissue Retractor
Designed by James L. Baldwin, MD

Sharp

#6312

Designed to hold back the fat pad andsoft tissues during total knee arthroplasty
The fenestrated paddle helps holds back the fat pad and soft tissues,
while the two long narrow prongs help penetrate the soft tissue, and rest
against the side of the tibia to help prevent rotation of the instrument.
PRODUCT NO:

6312 [Sharp Prongs]

Blunt

#6313

Overall Length: 9.875" (25,1 cm)
Pad Dimensions: 38 mm x 15 mm
Prong Depth: 22 mm

6313 [Blunt Prongs]

Overall Length: 9.75" (24,8 cm)
Pad Dimensions: 38 mm x 15 mm
Prong Depth: 20 mm

Modified Short Tip
Fat Pad Retractors
Designed by Robert Wubben, MD, withmodification by Mojieb Manzary, MD, FRCSC

Designed to help with soft
tissue and fat pad retraction in
the smaller knee, the blunted,
shortened end of the pointed
keel helps provide protection
against bony perforation

Small#3217-SPRODUCT NO'S:

3217-L [Large]

Overall Length: 9.625” (24,4 cm)
Prong Length: 12 mm

Large

3217-S [Small]#3217-L

Overall Length: 9.625” (24,4 cm)
Prong Length: 12 mm

Modified TKA Retractor Set
Designed by Robert Wubben, MD, with modification by David Ott, MD
PRODUCT NO’S:

3219-00 [Set]Right

Also available individually:#3219-R

3219-L [Left]

Overall Length: 10" (25,4 cm)
Prong Length: 20 mm

3219-R [Right]

Overall Length: 10" (25,4 cm)
Prong Length: 20 mm

Left#3219-L

Designed for soft tissue retraction,
the reduced phalange allows forease of placement in the lateralgutter, and helps avoid contact
with the lateral condyle

Wubben LateralFat Pad Retractor for TKRPRODUCT NO:

3218

Overall Length: 10" (25,4 cm)
Blade Width: 41 mm

16

Designed by Robert Wubben, MD

Designed to hold soft tissues
when inserting the TKR

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Multi-Purpose Knee
& Hip Retractors

PRODUCT NO’S:

4554-L [Left]

Overall Length: 11.25" (28,6 cm)
Blade Width: 1.5" (38 mm)

4554-R [Right]

Designed by Vasilios Mathews, MD

Overall Length: 11.25" (28,6 cm)
Blade Width: 1.5" (38 mm)

In knee surgery, the retractors
can be used to help protect the patellar tendon
behind the fin at the lateral tibial border. Also useful asa soft-tissue and fat pad retractor during prosthesis implantation,
helping to ensure a dry cancellous bed for cementation.
During direct anterior hip arthroplasty procedures, the fin of thisretractor fits the contours of the acetabular rim and retracts theanterior soft tissues, while the short length of the spike helps limit
the penetration into the neurovascular zones.

Right

Left

#4554-L

Wide Rake Retractors
with Ergonomic Handle
Designed for general use
soft tissue retraction,
the ergonomic handle
allows for a better grip
and less fatigue
Non-glare finish featured on
the metal retractor parts.

Designed for use
in both knee andhip arthroplasty
procedures#4554-R

Deep, Sharp #6051
Deep, Blunt #6052

PRODUCT NO’S:

6051 [Deep, Sharp]

6053 [Shallow, Sharp]

6052 [Deep, Blunt]6054 [Shallow, Blunt]

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 2.75" (7 cm)
Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 2.75" (7 cm)

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 1.875" (4,8 cm)

Shallow, Sharp #6053
Shallow, Blunt #6054

Sharp and dull teeth
available in two depths

Overall Length: 11.175 (28,3 cm)
Blade Width: 2.375" (6 cm)
Blade Depth: 1.875" (4,8 cm)

PRODUCT NO’S:

75 x 25 mm#6243

75 x 15 mm#6242

50 x 16 mm

Meyerding Type Retractors
with Ergonomic Handle
Designed for general use soft
tissue retraction, the ergonomic
handle allows for a better grip
and less fatigue

6241 [50 x 16 mm]

Overall Length: 8.875" (22,5 cm)
Blade Width: 16 mm
Blade Depth: 50 mm

6242 [75 x 15 mm]

Overall Length: 9" (22,9 cm)
Blade Width: 15 mm
Blade Depth: 75 mm

6243 [75 x 25 mm]

Overall Length: 9" (22,9 cm)
Blade Width: 25 mm
Blade Depth: 75 mm

Non-glare finish featured on
the metal retractor parts.

#6241

Rake Retractors
with Ergonomic Handle
Designed for general use
soft tissue retraction

PRODUCT NO’S:

4839 [3-Prong]

Overall Length: 9.5" (24,1 cm)
Rake Width: 13 mm
Rake Depth: 14 mm

3-Prong#4839

4840 [4-Prong]

Overall Length: 9.5" (24,1 cm)
Rake Width: 19 mm
Rake Depth: 14 mm

4-Prong#4840

1.800.548.2362       APRIL 2024       KNEE EXPOSURE & POSITIONING INSTRUMENTS

17

90° Bone Hook

Designed by Charles Taunt, DO

Designed to ergonomically help thesurgical assistant elevate the proximal
femur during TKA, the bone hook aids
the surgeon in accessing posterior
osteophytes and in applying local
anesthetic to the posterior capsuleTakes the place of an intramedullary device
when the IM canal has not been opened(robotic assistance) or when damaged orosteopenic bone is of concern.

PRODUCT NO’S:

5940-B [Blunt Tip]

Curve Diameter: 50 mm
Hook Depth: 6.5" (16,5 cm)
Handle Length: 5" (12,7 cm)

Blunt Tip

5940-S [Sharp Tip]

#5940-B

Curve Diameter: 50 mm
Hook Depth: 6.5" (16,5 cm)
Handle Length: 5" (12,7 cm)

Sharp Tip

#5940-S

Distal Femur Distractor

Upward Bent Handle

Helps distract the distal femur away from the proximal tibia#4220-01

Designed to distract the distal femur away from the proximal tibia during total
knee surgery. It is inserted into a pre-drilled hole in the distal femur. The
bent handle allows the femur to be distracted away fromthe tibia. The intramedullary rod portion is fluted.

Standard Handle#4220-00

PRODUCT NO’S:

4220-00 [Standard Handle]

Overall Length: 12.75" (32,4 cm)
Rod Offset from Handle: 4.5" (11,4 cm)

4220-01 [Upward Bent Handle]

Overall Length: 17.5" (49,6 cm)
Rod Length from Bend: 12.75" (32,4 cm)
Rod Offset from Handle: 4.5" (11,4 cm)

Gelpi Retractors
Standard

Ergonomic Handle

#4180

#4181

PRODUCT NO’S:

4180 [Standard]

Overall Length: 7.75" (19,7 cm)

4181 [With Ergonomic Handle]
Overall Length: 8.75" (22,2 cm)

Deep Meyerding Retractor
with Ergonomic Handle

A self-retaining soft tissue retractor for
use in hip, knee, and shoulder surgery
PRODUCT NO:

6244

Overall Length: 8.5" (21,6 cm)
Blade Depth: 3.5" (8,9 cm)
Blade Width: 1" (2,54 cm)

18

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYINNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Small 1.25" Depth#4191

Flat Gelpi
Retractors

Medium 1.75" Depth#4192

Designed to help
retract a broader areaof soft tissue or muscle

Large 3" Depth#4193

Deep 5" Depth

PRODUCT NO'S:#4194

4191 [Small]

Overall Length: 6.5" (16,5 cm)
Prong Depth: 1.25" (3,2 cm)

4192 [Medium]

Overall Length: 7.25" (18,4 cm)
Prong Depth: 1.75" (4,4 cm)

4193 [Large]

Overall Length: 9" (22,9 cm)
Prong Depth: 3" (7,6 cm)

4194 [Deep]

Overall Length: 10" (24,4 cm)
Prong Depth: 5" (12,7 cm)

The two largest sizes feature
an ergonomic handle forincreased comfort and control.

Ortho Self-Retaining Retractors

Helps separate the femur and tibia during knee
replacement procedures, where the calibrated
design can help to balance ligaments
Calibrated ratchet (in mm) is helpful in ligament balancing

Features a no-teeth design, available with flat orserrated outside blades
Can also be used for spine surgery where thecalibrated ratchet can be used to help accurately
measure the size of an opening – useful inprocedures to help assess bone graft needs
Also useful in foot & ankle surgery

Standard Handle#1842
Serrated Outside Pads

1842 [Small Flat]

1842-01 [Small Serrated]

1843 [Medium Flat]

1843-01 [Medium Serrated]

Overall Length: 9.25" (23,5 cm)
Blade Width: 10 mm
Blade Thickness: 1,7 mm#1843

Small Flat

PRODUCT NO’S:
Flat Outside Pads

Overall Length: 6.5" (16,5 cm)
Blade Width: 7 mm
Blade Thickness: 1,7 mm

Medium Flat

Overall Length: 6.5" (16,5 cm)
Blade Width: 7 mm
Blade Thickness: 1,7 mm

Medium Serrated#1843-01

Overall Length: 9.25" (23,5 cm)
Blade Width: 10 mm
Blade Thickness: 1,7 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Small Serrated#1842-01

Small Grip Handle
PRODUCT NO’S:
Serrated Outside Pads

1842-01-SG [Small Serrated, Small Grip]
Overall Length: 6.5" (16,5 cm)
Blade Width: 7 mm
Blade Thickness: 1.68 mm

Designed with the grip
closer together for easiergripping and to help
reduce hand fatigue

1.800.548.2362       APRIL 2024   

Small Serrated with Small Grip
#1842-01-SG    KNEE EXPOSURE & POSITIONING INSTRUMENTS

19

Lombardi Gap Balancing
Femoral Tibial Spreader

Parallel
at 20 mm

Designed by Adolph V. Lombardi Jr., MD

Locking mechanism modification designed by Munish C. Gupta, MD

Designed to help separate the femur
and tibia during total knee procedures,with the pads being parallel whenmeasured at 20 mm of separation
All models and sizes feature a calibrated handle to help
accurately gauge the gap, and makes it possible for two
spreaders to be used to assist in balancing ligaments.

Ratcheting Mechanism

Standard Handle

Large – Horizontal Grooved Pads #1878
Large – Diamond Cut Pads #1878-D

PRODUCT NO’S:

1878 [Large – Grooved Pads]
Overall Length: 9.25" (23,5 cm)
Pads: 22 mm x 13 mm
Opens to: 50 mm

1877 [Small – Grooved Pads]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

Ratcheting Mechanism

1878-D [Large – Diamond Cut Pads]
Overall Length: 9.25" (23,5 cm)
Pads: 22 mm x 13 mm
Opens to: 50 mm

Small – Horizontal Grooved Pads #1877
Small – Diamond Cut Pads #1877-D

1877-D [Small – Diamond Cut Pads]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Locking Mechanism
Grooved Pads

Diamond Cut Pads

Large – Horizontal Grooved Pads #1878-LR

Locking Mechanism
PRODUCT NO’S:

1878-LR [Large – Grooved Pads
with Locking Mechanism]

Locking Mechanism

Small – Horizontal Grooved Pads #1877-LR

Overall Length: 9.625" (24,4 cm)
Pads: 22 mm x 13 mm
Opens to: 45 mm

1877-LR [Small – Grooved Pads
with Locking Mechanism]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

1877-SP [Small – Smooth Pads
with Locking Mechanism]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

Lockingratchetmechanism
helps prevent
accidental
release, andprovides forcontrolled
adjustment
and easy
release.

Locking Mechanism

Small – Smooth Pads #1877-SP
Design modified by Mojieb Manzary, MD, FRCSC

Small Grip Handle

Designed with the grip closer together for easiergripping and to help reduce hand fatigue

PRODUCT NO:

1877-SG [Small with Small Gripand Grooved Pads
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

20

Ratcheting Mechanism

Small with Small Grip & Horizontal
Grooved Pads #1877-SG

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Lombardi Femoral
Tibial Spreader
Designed by Adolph V. Lombardi Jr., MD

Locking mechanism modification designed by Munish C. Gupta, MD

Thin pads help to separate the femur
and tibia during total knee procedures
All models and sizes feature a calibrated handle to help
accurately gauge the gap, and makes it possible for two
spreaders to be used to assist in balancing ligaments.

Standard Handle
PRODUCT NO’S:

Ratcheting Mechanism

1875 [Large – Grooved Pads]
Overall Length: 9.25" (23,5 cm)
Pads: 22 mm x 13 mm
Opens to: 50 mm

Large – Horizontal Grooved Pads #1875
Large – Diamond Cut Pads #1875-D

1875-D [Large – Diamond Cut Pads]
Overall Length: 9.25" (23,5 cm)
Pads: 22 mm x 13 mm
Opens to: 50 mm

1876 [Small – Grooved Pads]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

Ratcheting Mechanism

1876-D [Small – Diamond Cut Pads]

Small – Horizontal Grooved Pads #1876
Small – Diamond Cut Pads #1876-D

Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Grooved Pads

Diamond Cut Pads

Locking Mechanism

Locking Mechanism

Large – Horizontal Grooved Pads #1875-LR

PRODUCT NO’S:

1875-LR [Large – Grooved Pads
with Locking Mechanism]

Locking ratchetmechanism
helps prevent
accidental release,and provides
for controlled
adjustment andeasy release.

Overall Length: 9.625" (24,4 cm)
Pads: 22 mm x 13 mm
Opens to: 45 mm

1876-LR [Small – Grooved Pads
with Locking Mechanism]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

Locking Mechanism

Small – Horizontal Grooved Pads #1876-LR

Designed with the grip closer together for easiergripping and to help reduce hand fatigue

Small Grip Handle
PRODUCT NO:

1876-SG [Small with Small Gripand Grooved Pads]

Ratcheting Mechanism

Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 35 mm

Small with Small Grip & Horizontal
Grooved Pads #1876-SG

Femoral Tibial Coated Spreader Bar
PRODUCT NO:

1820

Overall Length: 13" (33 cm)
Coated Surface: 4" (10,2 cm)
Blade Width: 13 mm

1.800.548.2362       APRIL 2024   

Designed to separate the femur and tibia
when implant components are in place

The end is coated to help protect fromscratching component surfaces.

    KNEE EXPOSURE & POSITIONING INSTRUMENTS

21

Scott Femoral Tibial Tensor/Spreaders
Designed by Richard Scott, MD

Used before determining femoral component rotation to help properly
tense the medial and lateral ligaments and help assure a stable,balanced flexion gap
Four Pad
Configurations
Available

Narrow Fixed Pads
#1995

Wide Fixed Pads*#1996

Original with narrowpads, designed to be
used before making thefemoral and tibial cuts

Wide Block Pads
#1997

Round Pads
#1998

Three new wide pad styles, designed for use after the cuts have been made

PRODUCT NO’S:

1995 [Narrow Fixed Pads]
Overall Length: 7" (17,8 cm)
Blade Width: 7 mm
Opens to: 40 mm

1996 * [Wide Fixed Pads]
Overall Length: 7" (17,8 cm)
Pads: 22 mm x 13 mm
Opens to: 40 mm

1997 [Wide Block Pads]
Overall Length: 7" (17,8 cm)
Pads: 23 mm x 12 mm
Opens to: 40 mm

1998 [Round Pads]

Overall Length: 7" (17,8 cm)
Pads: 25 mm x 25 mm
Opens to: 40 mm

Surgical Technique available on our website.

US Patent #8,162,951 B2

Designed toprotect thesurface ofthe tibia*Pad Modification designed by Raymond H. Kim, MD

Sorrells Tibia Protector Plates
Designed by R. Barry Sorrells, MD

Small
65 x 40 mm
#1135

22

An important part of surgical technique during total knee arthroplasty
is the establishment of a symmetric balanced flexion gap. This canbe achieved by tensing the medial and lateral ligaments with laminar
spreaders and rotating the femoral component until a rectangularspace is formed. The calibrated Tensor/Spreader allows the surgeon
to choose a reproducible amount of tension across the medial orlateral flexion space.
In the varus knee, any medial release necessary to balance
the knee in extension is performed first. In the valgus knee, theflexion gap can be balanced before the extension gap if the lateralretinaculum (not the lateral collateral ligament) is all that needs
releasing to correct the deformity.
The spreader can be used before or after tibial preparation andalso during revision surgery after a well-aligned tibial platform has
been established.
The knee is flexed 90 degrees. Any medial retractor is removed.
The medial gap is tensed with a plain or calibrated laminar spreader
that is opened until the medial collateral ligament is fully tensed.
The calibrated tensor is applied laterally and opened to the desired
tension on the indicator. The femoral component is rotated until a
rectangular gap is formed based off the tibial cut or an external tibialalignment device (if the tibial resection has not yet been performed).

INNOMED.NET   PRODUCT NO’S:

1130 [Large] 75 mm x 45 mm
1135 [Small] 65 mm x 40 mm

Large
75 x 45 mm#1130   FREE TRIAL ON MOST INSTRUMENTS

Calibrated Femoral Tibial Spreaders

Locking Mechanism Version

Helps prevent accidental release, and provides
for controlled adjustment and easy release.

Helps separate the femur and tibia
during total knee replacement surgery

Speed Lock Version

Helps allow precise control and prevent unintended release.

Designed to remain in position, with the femur and tibia separated, without
the need of an assistant, and to minimize crushing the bone, even if
osteoporotic. A wide unobstructed view of the posterior compartment ispossible. Osteophytes on the posterior condyles of the femur and tibia can beseen and removed. The calibrated handle of the spreader makes it possible
for two spreaders to be used to assist the surgeon in balancing ligaments.

Coated Pad Version

Helps protect component surfaces when implants
are in place, and are slightly contoured to add stability
against the curved articulating implant surfaces.

Small Grip Handle

Small – 7"

Designed with the grip closer together
for easier gripping and to help reduce hand fatigue.
PRODUCT NO’S:

SMALL WITH STANDARD GRIP
1850 [Grooved Pads]
Overall Length: 7" (17,8 cm)
Pads: 23 x 12 mm
Opens to: 39 mm

1850-D [Diamond Cut Pads]
Overall Length: 7" (17,8 cm)
Pads: 23 x 12 mm
Opens to: 39 mm

Standard Grip

Small with Grooved Pads #1850
Small with Diamond Cut Pads #1850-D

Small with
Coated Pads

Small withRound Pads

#1850-01#1865

1850-01 [Coated Pads]
Overall Length: 7" (17,8 cm)
Pads: 18 x 15 mm
Opens to: 39 mm

Small – 7"1865 [Round Pads]

Overall Length: 7" (17,8 cm)
Pads: 25 x 25 mm
Opens to: 39 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYPRODUCT NO’S:

SMALL WITH SMALL GRIP HANDLE
1850-SG [Grooved Pads]
Overall Length: 7" (17,8 cm)
Pads: 23 x 12 mm
Opens to: 39 mm#1865-SG#1850-01-SG
Small Grip Handle

1850-01-SG [Coated Pads]

Small Grip
and Round Pads

Small Grip
and Coated Pads

Overall Length: 7" (17,8 cm)
Pads: 18 x 15 mm
Opens to: 39 mm

Small Gripand Grooved Pads

Designed with the grip closer together
for easier gripping and to help reduce hand fatigue#1850-SG

1865-SG [Round Pads]
Overall Length: 7" (17,8 cm)
Pads: 25 x 25 mm
Opens to: 39 mm

Small – 7"

PRODUCT NO’S:

SMALL WITH LOCKING MECHANISM
1850-LR [Grooved Pads]
Overall Length: 7" (17,8 cm)
Pads: 23 x 12 mm
Opens to: 39 mm

Locking Mechanism

Small with Grooved Pads
and Locking Mechanism

Small with Round Pads
and Locking Mechanism#1850-LR#1865-LR

1865-LR [Round Pads]
Overall Length: 7" (17,8 cm)
Pads: 25 x 25 mm
Opens to: 39 mm

Medium – 10"

PRODUCT NO’S:MEDIUM
1855 [Grooved Pads]

Overall Length: 10" (25,4 cm)
Pads: 23 x 14 mm
Opens to 50 mm

1855-D [Diamond Cut Pads]
Overall Length: 10" (25,4 cm)
Pads: 23 x 14 mm
Opens to 50 mm

1855-SL [Speed Lock* & Grooved
Pads]
Overall Length: 10" (25,4 cm)
Pads: 23 x 14 mm
Opens to 50 mm

Standard Grip

Medium with Grooved Pads #1855
Medium with Diamond Cut Pads #1855-D

Medium withRound Pads

Medium with Speed Lock
& Grooved Pads (No Calibrations)#1866#1855-SL1866 [Round Pads]

Overall Length: 10" (25,4 cm)
Pads: 25 x 25 mm
Opens to 50 mm

Large – 12"

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY*Speed lock modification designed by Nasim A. Rana, MD
PRODUCT NO:

LARGE1860 [Grooved Pads]

Overall Length: 12" (30,5 cm)
Pads: 25 x 16 mm
Opens to 65 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Grooved Pads

1.800.548.2362       APRIL 2024   

Diamond Cut Pads

Coated Pads

Round Pads

Large with
Grooved Pads
#1860    KNEE EXPOSURE & POSITIONING INSTRUMENTS

23

Robb Leg Positioner
Designed by William Robb, MD

Provides stable positioningof the knee during surgery

Slotted base allows the leg to be easily flexed or extended
during knee surgery
Slots are also designed to allow the foot piece to be rotated
The complete unit is steam and gas sterilizable
Supplied with a sterilizable table clamp which can beclamped over the sterile drape to the O.R. table side bar
Three (3) Sterile Pads/Wraps are included with each newpurchase (See below for Pad/Wrap information)

PRODUCT NO’S:

Base Dimensions: 21" x 11" (53,4cm x 27,9cm)

2630-11 [Robb Leg Positioner]
Optional & Replacement Parts:

2630-FP [Aluminum Footpiece Only]
2629-00 [Case of 10 Sterile Pads/Wraps]
2595 [Table Clamp]

Leg Positioner
Sterile Protective Pad & Wrap

Disposable, latex-free sterile foam padand cohesive wrap helps protect patientfrom pressure sores, abrasions andpossible neurological impairment
while securing foot into the bootCompatible with the Innomed’s Stulberg, Robb and Lombardi Leg Positioners

PRODUCT NO:

2629-00 [Case of 10 Sets]
2629-L [1 Set]

Each set includes 1 Pad and 1 Wrap.

24

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Stulberg Leg Positioner

PRODUCT NO’S:

Base Dimensions: 20" x 10.5" (50,8 cm x 26,7 cm)

2620-10 [Stulberg Leg Positioner]
Optional & Replacement Parts:

Designed by S. David Stulberg, MD

2620-FP [Aluminum Footpiece Only]
2629-00 [Case of 10 Sterile Pads/Wraps]

Provides stable positioning of the knee during surgery

Usage guide available on our website.

Allows the leg to be manipulated into the desired position and securely locked in place
Includes the necessary adjustments to tilt, rotate, and flex or extend the knee
Extension/flexion adjustments can be made with the quick releaseof the ratchet
In use, the base plate is clamped onto the operating table withthe vertical side bar, the base plate is draped, and the sterile
support plate lowered into the base plate
The patient’s foot is wrapped into the foot support with a sterile
bandage (additional padding may be used for thin tibias)
The complete unit is steam and gas sterilizable
Three (3) Sterile Pads/Wraps are included with eachnew purchase (See page 24 for Pad/Wrap information)

Tilt Bar

Loosening the Tilt Bar allows the knee to be tilted in either
direction. Tightening the bar locks the Yoke System in thedesired position.

Foot/Ankle Support

RatchetSupport Plate

Base Plate

The Ratchet allows the Yoke Assembly to be moved in a
precise gradual manner, the length of the Track. For fasteradjustments, downward pressure on the Ratchet Handle
releases the Yoke Assembly which then can be easily slid
the length of the Track.

Rotation Bar

Loosening the Rotation Bar allows the knee to be rotatedin either direction. Tightening the bar locks the Yoke
System in the desired direction.

1.800.548.2362       APRIL 2024       KNEE EXPOSURE & POSITIONING INSTRUMENTS

25

Chandran Thigh Lift Positioner

Post Screw

Set #4167-00#4150-PS

Designed by Rama Chandran, MD

Designed to help lift and position thethigh from above during knee surgery
Post Screw#4150-PS

Cradle #4167-01

PRODUCT NO’S:

4167-00 [Positioner Set]
Set Includes:

4167-01 [Cradle]

Dimensions: 7.5" x 7.75" (19,1 x 19,7 cm)

4167-02 [Rod]Width: 13.6" (34,5 cm)
Height: 20.6" (52,3 cm)

Rod #4167-02

Orange Strap

4150-PS [Post Screw]

Two included in set, one with this product number

2760-P [Silicone Pad]

Dimensions: 9” x 8” x ½” thick      (22,9 x 20,3 x 1,3 cm)[Orange Strap]

Dimensions: 2" x 31.5" (5,1 x 80 cm)
Two included in set, see below for replacement 10-pack

PRODUCT NO’S:
Optional Parts:

Optional Adapter #4167-03

4167-03 [Adapter]

Width: 12.4" (31,5 cm)
Height: 4.875" (12,4 cm)

9125 [Rotating Table Clamp]

Optional Adapter #4167-03
Optional Rotating
Table Clamp #9125

The optional
thigh lift adapter
is designed for use
with a hydraulic lift
device instead of themanual lift rod withtable clamp.

PRODUCT NO’S:
Replacement Parts:

8130-P [Orange Straps - 10-Pack]

Dimensions (Each): 2" x 31.5" (5,1 x 80 cm)

2760-P [Silicone Pad]

Dimensions: 9” x 8” x ½” thick      (22,9 x 20,3 x 1,3 cm)

Adjustable Knee & Tibial Positioner
Designed by Ashutosh Chaudhari, MD

Adjustable design allows for use in procedures around theknee such as tibial nailing, tibial condyle plating, patella
fracture fixation, supracondylar fracture plating, supracondylar
fracture nailing,and total knee replacement
PRODUCT NO’S:

2770-00 [Set]

Includes Positioner, Pad,
and Two Short Straps

Individual / Replacement Parts

2770-01 [Positioner]

Overall Length (Folded): 28" (71,1 cm)
Overall Length (Flat): 54.75" (139 cm)
Maximum Triangle Height: 14" (35,6 cm)
Width: 5.5" (14 cm)
Thickness (Folded): 1.8" (4,6 cm)
Thickness (Flat): .75" (1,9 cm)

2770-P [Silicone Pad]

Dimensions: 12" x 5.5" (30,5 cm X 14 cm)

2590-S [Short Straps] Pkg of 10

26

INNOMED.NET   Radiolucent. Steam sterilizable.

   FREE TRIAL ON MOST INSTRUMENTS

Lombardi Leg Positioner
Designed by Adolph V. Lombardi Jr., MD

Designed to hold the leg during total
knee surgery, the unrestricted design
helps allow for manipulation of the leg

Profile View

The footpiece consists of an open topped boot, separated
with four spacers and attached to a boot support plate.
The support plate is bent beyond the rear of the boot to help
provide support in different positions of flexion/extension.
The spacers allow for room to clean between the boot andsupport plate.
Two (2) Sterile Pads/Wraps are included with each purchase.
PRODUCT NO’S:

2622 [Lombardi Leg Positioner]

Base Length: 13" (33 cm)
Base Width: 7" tapers to 4" (17,8 x 10,2 cm)
Overall Height: 12.75" (32,4 cm)

Replacement Parts:

2629-00 [Sterile Pad & Wrap – Case of 10 Sets]

Kirschenbaum Foot Positioner
Designed by Ira Kirschenbaum, MD

Helps eliminate the use of sand bags
under the drape during total knee surgery

The foot rest is dome shaped for optimal foot contact andpositioning the leg in flexion, and can be rotated. The unit canbe used under the drape by attaching it to a standard tableattachment or it can be sterilized for use on top of the drape. It
can be attached to the table with the optional sterilizable tableclamp. Supplied with a removable, sterilizable silicone foot pad.

Long #2590

Short #2591

PRODUCT NO’S:

2590 [Foot Positioner – Long] 15.5" x 6" (39,4 cm x 15,2 cm)
2591 [Foot Positioner – Short] 9.5" x 6" (24,1 cm x 15,2 cm)
Optional & Replacement Parts:

2590-P [Large Replacement Pad] 16" x 9" (40,7 cm x 22,9 cm)
2591-P [Small Replacement Pad] 9.5" x 9.25" (24,1 cm x 23,5 cm)
2595 [Optional Table Clamp]

Stulberg Sliding Bolster
Designed by S. David Stulberg, MD

Helps eliminate the need for asand bag during total knee surgery
The base plate is attached to the table and thesterile sliding bolster is placed on top of thesterile drape. The bolster can be adjusted for
different angles of knee flexion during surgery.
PRODUCT NO’S:

2730

Base Dimensions:20" x 10.5" (50,8 cm x 26,7 cm)

1.800.548.2362       APRIL 2024       KNEE EXPOSURE & POSITIONING INSTRUMENTS

27

Use in Knee Surgery

Use in Hip Surgery
PRODUCT NO’S:

2750-00 [Set]

Berger Block Positioner

Dimensions with Pads: 4.75" x 6.75" x 8"
(12,1 x 17,1 x 20,3 cm)

Set Includes / Available Individually:

Designed by Richard Berger, MD

2750-01 [Positioner Only]

Designed for lower
extremity positioning
with dual height options

Dimensions: 4.125" x 6.125" x 8"
(10,5 x 15,6 x 20,3 cm)

2750-P [Positioner Pad Only][Brown Strap Only] (2) Included in Set
Optional Items:

2750-S [Brown Straps] Pkg of 10

Sanders Extremity Positioning Tubes
Designed to support the knee and ankle
during lower extremity surgery
Designed by Richard A. Sanders, MD

The 4" (10,2 cm) tube elevates the foot and ankle for ankle
fracture surgery.
The 6" (15,2 cm) tube lifts the knee off the operating table
and allows for approximately 30° of knee flexion. Very useful for
closure of total knee incisions, supporting fractures of the distal
femur, and tibia plateau fractures.
The tubes are made of aluminum, allowing them to be
autoclaved. They help eliminate the need for rolled sheet bolsters.
PRODUCT NO’S:

2740-01 [Small]

Diameter: 4" (10,2 cm)
Width: 8" (20,3 cm)

2740-02 [Large]

Diameter: 6" (15,2 cm)
Width: 8" (20,3 cm)

4" #2740-01

6" #2740-02

Lower Extremity Leg Positioner
Designed by Ronald Romanelli, MD

Used to support knee and leg during
surgery, and can be used for casting
Utilized for rodding of femurs or tibias
Also useful for knee surgery and closures
Very supportive, distributes stresses on leg,
used instead of bolsters
Supplied with one autoclavable silicone pad
Aluminum positioner is radiolucent and gas
or steam sterilizable
PRODUCT NO’S:

2745

Dimensions: 5.5" x 9.5" x 9.25" (14 x 24 x 23,5 cm)

Replacement Parts:

2760-P [Silicone Pad]

28

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

PRODUCT NO’S:

2589-00 [Complete Assembly]

Hyperflex Foot
Positioner Assembly

Overall Length: 19" (48,3 cm)

Individual/Replacement Parts:

2589-01 [Foot Positioner]
Overall Length: 18" (45,7 cm)

2590-B [Positioner Bar]

Designed by Morteza Meftah, MD and Ira Kirschenbaum, MD

Overall Length: 19" (48,3 cm)

Designed to help secure thefoot for positioning of the kneein the hyperflex position

2730-P [Pad & Two Straps]
4150-PS [Post Screw]
Optional Parts:

2590-S [Black Straps] Pkg of 10

Cherf Leg Holder

Designed by John Cherf, MD

Supports the lower extremity forprepping before knee or hip surgery
Useful for all lower extremity procedures and isparticularly helpful for supporting the leg with thepatient positioned in the lateral position. By holding
the foot/ankle in an externally rotated position,
the knee can be locked into extension which helps
eliminate the need for manual support.
May also be used to support the limb for surgicalpatients in the supine position such as for knee
and foot/ankle procedures.
PRODUCT NO’S:

2270
Replacement Parts:

4150-PD3 [Set of 3 Small Pads]

Patient Self Stress Assembly Set
Designed by Kyle Cook, RTR and David Mauerhan, MD

Designed to help
position a patient for
X-ray evaluation to help
determine candidacy
for Unicondylar Knee
Arthroplasty

Contoured Cube

#2741-02

Triangle Positioner#2741-01

PRODUCT NO'S:

2741-00 [Set]
Individual Instruments:

2741-01 [Triangle Positioner]

Dimensions: 24" x 9" x 9" (61 cm x 23 cm x 23 cm)

2741-02 [Contoured Cube]

Dimensions: 11" x 9" x 6" (28 cm x 23 cm x 15,2 cm)

1.800.548.2362       APRIL 2024       KNEE EXPOSURE & POSITIONING INSTRUMENTS

29

Modified 90°
Leg Stabilizer
Designed by Gregory Fanelli, MD

Useful in total knee surgery
to hold the leg in position
Helps to open up the knee joint whenpressure is applied to the lower leg. Pad
and sterilizable table clamp included.
PRODUCT NO:

2725

Post Height: 11.375" (28,9 cm)
Pad Length: 9" (22,9 cm)
Pad Diameter: 3" (7,6 cm)

Replacement Parts:

9120 [Table Clamp]
8840-P [Pad]

Leg Stabilizer
Designed by Gregory Fanelli, MD

Useful in arthroscopic
knee surgery to holdthe leg in position
Helps to open up the knee joint whenpressure is applied to the lower leg.
Sterilizable table clamp included.
PRODUCT NO:

8840

Overall Length: 18.5" (47 cm)
Handle Length: 9.25" (23,5 cm)
Pad Diameter: 3" (7,6 cm)

Replacement Parts:

9120 [Table Clamp]
8840-P [Pad]Durham Leg Positioner

Designed by Al Durham, MD

Placed against the thigh, helping to holdthe leg upright in knee surgery
Supplied with a sterilizable table clamp. The pad is madeof semi-dense foam to help prevent pressure points andis sealed with a washable coating.

PRODUCT NO:

4105
Replacement Parts:

9120 [Table Clamp]
4105-P [Pad]30

INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Stanton Arthroscopic Leg Holder

Designed by John Stanton, MD

Designed to securely hold legs of various sizes for arthroscopic surgery
Sliding leg holder can be adjusted for small calves or to accommodate large thighs
Locking pin prevents sides from spreading apart
Strap can be placed high or low through the slots in the side plates toaccommodate large/small limbs
Strap is strongly secured with a toothed clamp
Support rod, when clamped into a standard table clamp,
helps to prevent rotation
PRODUCT NO’S:

PROTOTYPE SHOWN

4045

Dimensions: 16.5" L x 8.5" H x 3.5" W      (42 cm L x 21,6 cm H x 9 cm W)
Fits Legs: From 4" to 11" (10 cm to 28 cm)

Replacement Parts

4045-S [Strap]

Overall Length: 28" (71,2 cm)

PROTOTYPE SHOWN

George Arthroscopic Knee Positioner
Designed by Michael S. George, MD

Provides lateral as well as superior support whichallows valgus stress to open the medial compartment
Shape does not squeeze the thigh, making the need for a thigh tourniquet
optional. If desired, the unit can easily be rotated out of the way without
disrupting the sterile field. Using with a standard operating table clamp,
the unit can easily be raised or lowered to accommodate all thigh sizes.
PRODUCT NO’S:

2735

Overall Height: 22" (55,9 cm)
Post Height: 12" (30,5 cm)
Pad Width: 3" (7,6 cm)

Replacement Parts

2735-P [Pad]Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

1.800.548.2362       APRIL 2024   

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:    KNEE EXPOSURE & POSITIONING INSTRUMENTS

31PRSRT. STD.
U.S. POSTAGE
PAID
BOURBON, IN

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 29

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

Fromm Femur & Tibia Triangles
Designed by S.E. Fromm, MD *Extra Small Triangle designed by S.E. Fromm, MD & Kenneth Merriman, MD

Used for femur and tibia positioning
during nailing, repairs and fractures
Designed to position and hold the femur and tibia during
intramedullary nailing of the tibia, ligament repairs andextremity fractures. Allows knee to be flexed greater than 90° toallow reaming and nail insertion without displacing fracture. The
triangles are available in four heights: 8.5", 11", 14", and 16".
The three smaller triangles are designed to fit inside the larger
triangle for storage. They are supplied with an autoclavablesilicone cushioning pad and Velcro* straps. The triangles arealso radiolucent and gas or steam sterilizable.
PRODUCT NO’S:

2760-00 [Set of 3] Angles: Top 30°, Two Bottom 75°
2760-01 [11"] Base: 6" (15,2 cm), Height: 11" (27,9 cm)
2760-02 [14"] Base: 7" (17,8 cm), Height: 14" (35,6 cm)
2760-03 [16"] Base: 9" (22,9 cm), Height: 16" (40,7 cm)

16” #2760-03

14” #2760-02

Tibia Reduced For:

RetrogradeFemoral Nailing11” #2760-01

Separately –8.5” #2760-XS Sold
Not In Set

Sold Separately – Not In Set:

2760-XS [8.5"] Base 5" (12,7 cm), Height: 8.5" (21,6 cm)
Replacement Parts:

2760-P [Silicone Pad]
2760-S [Straps] Package of 18 – 6 Blue / 12 Green
8100-P [Green Straps for Femur, Long] Pkg of 10
8120-P [Blue Straps for Tibia, Short] Package of 10
8120-SP [Straps for 2760-XS] Package of 10*Velcro® is a registered trademark of the Velcro Companies.

INNOMED, INC

103 Estus Drive
Savannah, GA 31404
Tel 912.236.0000
Fax 912.236.7766

ISO 13485:2016

innomed.net
info@innomed.net• Open Reduction and Internal Fixation (ORIF)• Application of uni- or multi-plane external fixator• Knee ligament repairs and/or reconstruction

Innomed-Europe LLC

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74
www.innomed-europe.com
orders@innomed-europe.com

Triangle holds femur reduced(prevents sagging)

RetrogradeFemoral Nailing

Tibial Nailing

Innomed-Europe GmbH

c/o Emons Logistik GmbHIn Rammelswiesen 9
D-78056 Villingen-Schwenningen, Germany
Tel 0049 (0) 7720 46110 60
www.innomed-europe.com
orders@innomed-europe.com

1.800.548.2362   INNOMED.NET

04
24© 2024 Innomed, Inc.
All Rights Reserved