Ultrasonic sensor
UMC3000-30H-E5-5M< Front of transducer and housing manufactured entirely fromstainless steel< Degree of protection IP68 / IP69K< Programmable via DTM with PACTWARE< Mounting bracket MH-30H-01 included in delivery

Single head system

Function
The enclosure and transducer of this ultrasonic sensor form a hermetically sealed unit. Therefore the sensor is suitable for all applications where avery high tightness is required. Since the sensor housing is made exclusively of V4A stainless steel and all seals are made of highly chemicalresistant materials, this sensor is also predestined for use in chemically aggressive environments.
For reliable operation, due to the special design of this sensor, solely the enclosed mounting accessories must be used.

Dimensions

ø 28

ø 30

ø 6.3

100

Release date: 2023-01-17 Date of issue: 2023-01-17 Filename: 70128216_eng.pdf

113.5

3.5

11

Technical Data
General specifications
Sensing range

200 ... 3000 mm

Adjustment range

240 ... 3000 mm

Dead band

0 ... 200 mm

Standard target plate

100 mm x 100 mm

Transducer frequency

approx. 100 kHz

Response delay≤ 200 msIndicators/operating means
LED green

Operating display

LED yellow

switching state

LED rederror

Electrical specifications

Refer to "General Notes Relating to Pepperl+Fuchs Product Information".
Pepperl+Fuchs Group
USA: +1 330 486 0001
Germany: +49 621 776 1111
fa-info@de.pepperl-fuchs.com
www.pepperl-fuchs.com
fa-info@us.pepperl-fuchs.com

Singapore: +65 6779 9091
fa-info@sg.pepperl-fuchs.com

1

Ultrasonic sensor

UMC3000-30H-E5-5M

Technical Data
Operating voltage

UB

10 ... 30 V DC

No-load supply current

I0≤ 50 mA

Input/Output
Input/output type

1 synchronization connection, bidirectional

0 Level0 ... 1 V

1 Level4 V ... UB

Input impedance> 12 kΩOutput rated operating current< 12 mA

Pulse length≥ 200 µs

Pulse interval≥ 2 msSynchronization frequency
Common mode operation≤ 20 HzMultiplex operation≤ 20/n Hz, n = number of sensors n ≤ 10 (factory setting: 5 )Input
Input type1 program input

Level (switch point 1)

0 ... 1 V

Level (switch point 2)

4 V ... UB

Input impedance> 10 kΩ

Pulse length

2 ... 5 sOutput
Output type1 switching output E5, PNP NO/NC, programmable

Rated operating current

Ie

200 mA , short-circuit/overload protected

Voltage dropUd≤2V

Repeat accuracy≤ 0.1 % of full-scale value

Switching frequencyf≤ 2.8 Hz

Range hysteresis

H

programmable , preset to 1 mm

Temperature influence< 1.5 % of full-scale value

Compliance with standards and directives
Standard conformity
Standards

EN IEC 60947-5-2:2020
IEC 60947-5-2:2019

Approvals and certificates
CCC approval

CCC approval / marking not required for products rated ≤36 V

Ambient conditions
Ambient temperature-25 ... 60 °C (-13 ... 140 °F)

Storage temperature-40 ... 85 °C (-40 ... 185 °F)

Release date: 2023-01-17 Date of issue: 2023-01-17 Filename: 70128216_eng.pdf

Mechanical specifications
Connection typecable PUR , 5 m

Core cross section

5 x 0.5 mm2

Housing diameter

30 mm

Degree of protection

IP68 / IP69K

Material
Housing

Stainless steel 1.4404 / AISI 316L
LED window: VMQ Elastosil LR 3003/Shore 50 A

Transducer

Stainless steel 1.4435 / AISI 316L

Mass425 g

Factory settings
Outputnear switch point: 240 mmfar switch point: 3000 mm
output function: Window mode
output behavior: NO contact

General information
Supplementary information

Switch settings of the external programming adapter:"output load": pull-down"output logic": inv

Refer to "General Notes Relating to Pepperl+Fuchs Product Information".
Pepperl+Fuchs Group
USA: +1 330 486 0001
Germany: +49 621 776 1111
fa-info@de.pepperl-fuchs.com
www.pepperl-fuchs.com
fa-info@us.pepperl-fuchs.com

Singapore: +65 6779 9091
fa-info@sg.pepperl-fuchs.com

2

Ultrasonic sensor

UMC3000-30H-E5-5M

ConnectionBNL+WH

Teach input

GY

Synchronization

BK

Switching output

BU

L-

Characteristic Curve
Characteristic response curve
Distance Y [m]
2.0
1.5
1.0
0.5
0.0-0.5-1.0-1.5-2.0012345Distance X [m]YXflat surface 100 mm x 100 mm
round bar, Ø 25 mm

Switching output modes
1. Window mode, normally open
A1 < A2:

A1

object distanceA2

2. Window mode, normally closed
A2 < A1:

A2

A1

3. One switch point, normally open
A1 -> ¥:A2
4. One switch point, normally closed
A2 -> ¥:A1

Release date: 2023-01-17 Date of issue: 2023-01-17 Filename: 70128216_eng.pdf

5. A1 -> ¥, A2 -> ¥: Object presence detection mode
Object detected: Switch output closed
No object detected: Switch output open

Accessories
UC-PROG1-USB

Programming adapter

V15S-G-0,3M-PURWAGO

Male cordset, M12, 5-pin, PUR cable with WAGO terminals

MH-30H-01

Mounting aid, 30 mm

Refer to "General Notes Relating to Pepperl+Fuchs Product Information".
Pepperl+Fuchs Group
USA: +1 330 486 0001
Germany: +49 621 776 1111
fa-info@de.pepperl-fuchs.com
www.pepperl-fuchs.com
fa-info@us.pepperl-fuchs.com

Singapore: +65 6779 9091
fa-info@sg.pepperl-fuchs.com

3Ultrasonic sensor

UMC3000-30H-E5-5M

Mounting
Mounting instructions
Comply with the minimum permissible bending radius of 70 mm, if you install the connecting cable!

The mounting accessories included with the sensor must be used in order to ensure reliable operation!

Commissioning
Adjustment possibilities
The sensor is equipped with 1 switching output with 2 programmable switch points. The programming of the switch points andthe of output behavior can be done in two different ways:• Using the teach input of the sensor• Using the sensor‘s serial interface. This method requires an external programming adapter and the corresponding software.
You will find the download link for the software at www.pepperl-fuchs.com on the product page of the sensor.
Synchronisation
The sensor features a synchronization input for suppressing ultrasonic mutual interference ("crosstalk“). The following
synchronization modes are available:
1. Automatic multiplex mode
2. Automatic master slave common mode
3. Externally controlled synchronization
Further Documentation

Release date: 2023-01-17 Date of issue: 2023-01-17 Filename: 70128216_eng.pdf

For information on programming and synchronisation you may refer to the commissioning instruction.

Refer to "General Notes Relating to Pepperl+Fuchs Product Information".
Pepperl+Fuchs Group
USA: +1 330 486 0001
Germany: +49 621 776 1111
www.pepperl-fuchs.com
fa-info@us.pepperl-fuchs.com
fa-info@de.pepperl-fuchs.com

Singapore: +65 6779 9091
fa-info@sg.pepperl-fuchs.com

4