J A JNUULAYR Y

2023

OR T HOPEDIC INS T RUMEN T S®

MantisScrewdriver Distractor
Page 21
Redler Percutaneous
Pin Clamp
Page 5

Ortho Mini Chisels & Gouges
Page 23

YezerskiSmall Bone Rongeur
Page 29

Kakar
Carpal Tunnel Retractors
Page 12

LawtonScrew
Extractors
Page 27

Hand & Wrist Instruments
1.800.548.2362 • INNOMED.NET

Auerbach Hand Positioner Set
Designed by David Auerbach MD

Designed to position and retract the skin for
surgical exposures of the hand, wrist and forearm

Hand Plate

Hand TrayPRODUCT NO’S:

1747-00[Auerbach Hand Positioner Set]

Also available individually:

1747-01[Hand Plate]1747-02[Hand Tray]

Thumb Post

Thumb Post Clip

Dimensions: 15" x 7" (38,1 x 17,8 cm)
Dimensions: 13.75" x 9.75" (34,9 x 24,8 cm)

1747-03 [Thumb Post]
1747-03-C [Thumb Post Clip]
1747-04 [Cord Clip]

Seven (7) included in Set, One (1) with this product number.

1747-05[Retractor]

1747-06[Wrist Strap Buckle]

1747-07[Wrist Strap]

Four (4) included in Set, One (1) with this product number.

Cord Clips (7)Two (2) included in Set, One (1) with this product number.
Two (2) included in Set, One (1) with this product number.

1747-08-6 [Set of 6 Cords]
1747-09 [Suction Holder]

Thumb Post & Clip

Retractors (4)Wrist Strap
Buckles (2)Suction Holder

Shown attached to plate

Suction Holder

Wrist Straps (2)Insert in any corner
to help remove blood
accumulating in tray

Cords (6)2

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

What’s New In This Catalog?
a snapshot of all the

instruments within
Roberts Pin Bending
CannulaSet
Page 25

Chandran Distal Biceps
TissueProtector
Page 32

MantisScrewdriver Distractor
Page 21

Pointed Fracture Reduction Clamps
Designed by Reza Firoozabadi, MD MA

Versatile set of fracture reduction clamps,
each with a specific tine design that allows
for appropriate vector placement so thatanatomic reduction can be obtained in a
number of different types of fractures

Straight Left & Right

Curved Left, Straight Right

1.9 mm tines allow for a snug fit in 2 mm drill holes
Tines angled to prevent clamp “slippage” with compression
Straight tines can be placed deep within bone which allows
for far cortex compression.
Clamps incorporate a box joint design that prevents clamp
joint loosening and the need for tightening.
Example applications: any transverse fracture (straight-straightclamp), both bone forearm fractures, olecranon fractures,
medial malleolus fractures, and many more.
Speed Lock Style: Extra-long spin down allows for increased
range of clamp use, and open-topped joint rotates to allow forincreased range of opening, and also allows for quick release

Small with
Speed Lock

PRODUCT NO’S:
SMALL WITH SPEED LOCK MECHANISM

Medium with
Speed Lock

Small with
Ratchet3666 [Straight Left & Right]
Overall Length: 5.5" (14 cm)

3667 [Curved Left & Right]
Overall Length: 5.5" (14 cm)

3666-L [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

3666-R [Straight Left, Curved Right]
Overall Length: 5.5" (14 cm)

MEDIUM WITH SPEED LOCK MECHANISM

3666-01 [Straight Left & Right]
Overall Length: 7" (17,8 cm)

Two styles — Speed Lock and Ratchet — each available in four tine configurations

3667-01 [Curved Left & Right]
Overall Length: 7" (17,8 cm)

3666-L-01 [Curved Left, Straight Right]
Overall Length: 7" (17,8 cm)

3666-R-01 [Straight Left, Curved Right]
Overall Length: 7" (17,8 cm)

SMALL WITH RATCHET MECHANISM

3668 [Straight Left & Right]
Overall Length: 5.5" (14 cm)

3669 [Curved Left & Right]
Overall Length: 5.5" (14 cm)

3668-L [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

3668-R [Straight Left, Curved Right]
Overall Length: 5.5" (14 cm)

Straight
Left & Right

1.800.548.2362       JULY 2023   CurvedLeft & Right   HAND & WRIST INSTRUMENTS

Curved Left,
Straight Right

Straight Left,
Curved Right3

Faillace Extra Small Bone Clamp
Designed by John J. Faillace, MD

Delicate enough to use on metacarpals but
strong enough for distal radius and larger
bones with its extra long ratchet
PRODUCT NO:

1171

Overall Length: 5" (12,7 cm)
Jaw Length: 1" (2,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Small Bone Holding Forceps
with Long Ratchet
Designed for use in stabilizationof a fracture or osteotomy
PRODUCT NO:

1170

Overall Length: 5.75" (14,6 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

OrthoLucent™
O’Brien Bone Clamp
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
The carbon fiber PEEK material is strong, lightweight,
completely radiolucent, can be steam sterilized, andhelps to prevent from marring component surfaces.
PRODUCT NO:

1815-R

Overall Length: 5.25" (13,3 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

O’Brien Bone Clamp
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
PRODUCT NO:

1816

Overall Length: 5.25" (13,3 cm)

4

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Redler Percutaneous Pin Clamp

Holds a small bone in apposition duringpercutaneous pinning of a fracture

Three Tube Sizes
Available

Designed with a proximal pin tube with teeth; the tube guides
the pin and the teeth help keep the tube in place on the bone.
The distal tip is used to control the bone fragment. Includes along ratchet for locking on various sized bones, from 1 mm to 14
mm. Also useful during insertion of cannulated screw guide wires.
PRODUCT NO’S:

Overall Length: 5" (12,7 cm)

1810-35 Tube Diameter: .035" (.9 mm)
1810-45* Tube Diameter: .045" (1.1 mm)
1810-62 Tube Diameter: .062" (1.6 mm)

MADE EXCLUSIVELYFOR INNOMED IN*

GERMANY
Designed by M.R. Redler, MD

Chang Pin Clamp
Designed by Win Chang, MD

Designed to allow accurate insertionof pins for internal fixation
Used for small bones, the clamp allows
accurate insertion of pins for internal fixation.
The cannula has a 1.8 mm internal diameter.
PRODUCT NO:

1760-01(Prototype used in
X-ray images)

Cannula Internal Diameter: 1.8 mm
Overall Length: 6" (15,2 cm)
Locking Ratchet Opens To: 25 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Lewin Small Bone Clamp
PRODUCT NO:

4685

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 5" (12,7 cm)

PRODUCT NO:

1803

Cannula Diameter: .062" (1.6 mm)
Overall Length: 5.25" (13,3 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYGERMANY

Bargo Bone Holding Clamp
Designed by Lonnie Bargo, CST/CFA

Designed to aid in the reduction of various fractures,and can help secure a plate in place during installation
Designed to aid in the reduction of various fractures such as:
spiral, transverse, compound, oblique, or butterfly. The clamp
can also be used to secure a plate in place while the screw
holes are being drilled and screws inserted. The fracture site
can also be manipulated with the clamp being used as alever. Teeth in the jaws allows for a better grip and a ratchet
locking handle allows use on various bone diameters.

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

5

Redler Wrist Bone Clamp with Wire Guide
Designed by M.R. Redler, MD

Designed to hold bony fragments inplace for placement of guide wires
Can be used for:Placement of pins across distal radius fracturesor across carpal bones
Arthroscopically assisted fixation in the wrist
Fracture fragments about the elbow
Placement of guide wires during
the open reduction and internal
fixation of a patella fracture

Two sizes available:
For use with 0.045" (1.1 mm)
or 0.062" (1.6 mm) K-wires.

PRODUCT NO’S:

1885-45

For Pins up to .045" (1,1 mm)
Overall Length: 9.5" (24,1 cm)
Jaw opens to: 3.5" (8,9 cm)

1885-62

For Pins up to .062" (1,6 mm)
Overall Length: 9.5" (24,1 cm)
Jaw opens to: 3.5" (8,9 cm)

Stanton Articulating
Small Bone Clamps
Designed by John L. Stanton, MD

Compress with a towel clamp

Opposing clamps
facilitate manipulationof fracture ends

The small tube allows use of a towel clamp
to compress non-union and shortening
osteotomies during fixation, as well as toallow the use of Gelpi retractors to distract
malunions during revision surgery.

PRODUCT NO’S:

1811-00 [Set of Two]
Also available individually:

1811-L [Left]

Overall Length: 5.125" (13 cm)
Curved Plate Radius: 5 mm
Pin Hole for Pins Up To: 2,4 mm

1811-R [Right]

Distract with a gelpi retractor

Overall Length: 5.125" (13 cm)
Curved Plate Radius: 5 mm
Pin Hole for Pins Up To: 2,4 mm

Bush Small Bone
Reduction Forceps
Designed by Andrew P. Bush, MD

Designed to help hold a smallbone or bone plate in position
for reduction and fixation
Opens to approximately .5” (13 mm).
PRODUCT NO’S:

1889 [Single]

Overall Length: 4.5” (11,4 cm)
Jaw Width: .15” (3,7 mm)

1888 [Double]

Overall Length: 4.5” (11,4 cm)
Jaw Width: .7” (17,7 mm)

6WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Chen Low Profile Plate/Bone Clamp

The rotating nut is used totighten/release the center arm
once in the desired position

Designed by Franklin Chen, MD

Designed for plate to bone clamping
in a variety of lower and upper
extremity fractures

The freely swivelingcenter arm allows foreasy placement, as well
as for quick release, aftergetting the legs in position

Useful for diaphyseal forearm fractures,
humerus fractures, and distal radius fractures.
PRODUCT NO:

1639

Overall Length: 4.5" (11,4 cm)
Prong Depth: 1" (2,5 cm)
End Prong Width: 6 mm|16 mm Gap|6 mm
Clamps from: .375" to 2.4" (1 to 6 cm)

Durham Bone Reduction Clamp

Large Window
Provides space to allow a boneplate to be slid into position
without removing the clamp

Designed by Alfred A. Durham, MD

Allows application of a bone plate
without removing the reduction clamp—
designed for medium size bones suchas the fibula, ulna, and radius
The wide window directly above the jaw provides
space to allow a bone plate to be slid into
position without removing the clamp.
PRODUCT NO:

3652

Overall Length: 7" (17,8 cm)

Mogul K-Wire/Pin Insertion Guide
Designed by Stuart J. Mogul, DPM, FACFAS

A guide designed for passing guidepins or k-wires through two adjacentmetatarsal bones
PRODUCT NO:

3017

Dimensions: 2.375" Tall x 3.75" Wide (6 x 9,5 cm)
Maximum Pin Diameter: 3/32" (2,4 mm)
Maximum Clamped Opening: 2" (5,1 cm)
Minimum Clamped Opening: .375" (1 cm)
Pin/K-Wire Guide Length: .925" (23,5 mm)

Argintar Claw Drill Guide Wire/Suture Passer
Designed by Evan Argintar MD

Expandable claw design allows for minimally invasive,
reproducible one-step wire/suture passage

PRODUCT NO:

8315-00 [Set: (1) Claw,(1) Wire/Suture Pin]
8315-01 [Claw Unit]

Especially helpful during applications where a suture will bepassed—particularly when soft tissue dissection is to beminimized, such as wrist reconstruction (DRUJ), elbow
reconstruction (ULCL/MCL), foot-ankle reconstruction(ATFL), quad/patella tendon repair surgery, andmulti-ligament knee reconstruction (MCL/LCL).

1.800.548.2362       JULY 2023   

Overall Dimensions: 2.5" x 4"–6"   (6,4 cm x 10,2 cm–15,2 cm)

1227 [3/32" (2 mm) Pin
with Wire/Suture Hole]
Overall Length: 6" (15,2 cm)   HAND & WRIST INSTRUMENTS

7

Resnick Allis Bone Clamp

Designed by Charles T. Resnick MD

A traditional Allis Bone Clamp
designed with a longer ratchet
which allows for a wider opening
to allow a bone to be clamped
and locked onto
PRODUCT NO:

1385

Overall Length: 6" (15,2 cm)
Ratcheted Clamp Opens to: 37 mm
Clamp End Width: 4.7 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Coated Allis Bone Clamps
Modification of design by Charles T. Resnick MD

A traditional Allis Bone Clamp
designed with a longer ratchet—for
a wider opening to allow a boneand plate to be clamped and locked
onto—and coated end(s) to preventfrom marring a component surface
PRODUCT NO’S:

1381 [One Coated End]

Overall Length: 6.125" (15,9 cm)
Ratcheted Clamp Opens to: 35 mm
Non-coated-end Width: 4 mm

1382 [Two Coated Ends]

Overall Length: 6.125" (15,9 cm)
Ratcheted Clamp Opens to: 35 mm
Non-coated-end Width: 4 mm

Slavitt Phalangeal Forceps

Designed by Jerome Slavitt, DPM

Designed to enable the surgeon to provide joint
distraction and stability during joint placement at
the base of the proximal phalanx of the lesser digits

PRODUCT NO:

1163

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 6" (15,2 cm)
Clamp Internal Opening Diameter: 4 mm

GERMANY

Helps to distract the joint and hold the bone, allowing easieraccess to the base. Can also be used for digital fusions tohold bones better for drilling and cutting applications.

Rudisill Locking
Small Bone Reduction Forcep
Designed by Ed Rudisill, MD

For reduction of hand phalanx
and metacarpal fractures
PRODUCT NO:

2017

Overall Length: 4.875" (12,4 cm)

8

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Ratcheting Reduction Clamp Kit
Designed by Michael Craig, OPA-C

Designed as a soft tissue sparing
fracture reduction clamp
PRODUCT NO’S:

3840-00 [Clamp Kit]
Also available Individually:

3840-02 [Plate Point]

Overall Length: 1" (2,54 cm)

3840-03 [Screw Point]

High torque can help provide bone and joint
reduction without squeezing surrounding tissues
Swivel points are placed on the bone, plate, or
screw and the ratcheting dial is turned to thedesired torque, allowing hands free operation
Swivel point design allows the clamp to be easily
moved from x-ray view without losing reduction
Screw Point fits into a screw head
Plate Point fits into a 3.5 mm plate hole

Overall Length: .875" (2,2 cm)

3840-04 [Percutaneous Point]

2 included in set, one with this product number
Overall Length: 1" (2,54 cm)

Plate
Point

3840-MA [Ratcheting Reduction
Mobile Arm with Ratchet Knob]

Screw
Point

Percutaneus
PointsOverall Length: 6.5" (16,5 cm)

3840-SA [Ratcheting Reduction
Stationary Arm]

Kit includes:(1) Ratcheting Reduction Stationary Arm,
(1) Ratcheting Reduction Mobile Arm with Ratchet Knob,(1) Plate Point, (1) Screw Point, and (2) Percutaneous Points

Overall Length: 10.5" (26,7 cm)
Width: 9" (22,9 cm)
Height: 6" (15,2 cm)

Resnick Small Bone Tamp with Oblique K-Wire Hole
Designed by Charles Resnick, MD

Design allows for the concurrent reduction of a fracture andplacement of a wire into the fracture site — especiallyhelpful when the surgical exposure is small andtight, the fracture fragments are small,and the reduction is demanding

TWO SIZES AVAILABLE:
Wire Hole for K-wires up
to 1.1 mm (.045”) or
1.6 mm (.062”)

BeforePRODUCT NO’S:

After

The serrated distal end minimizes slippage on the cortical surface, does notinterfere with the placement of the guidewire and allows for subsequent surgeondecided, intraoperative angulation of the wiring once the first cortex is drilled
Especially useful in fractures where there is involvement of an articular surface,for example, mallet fractures of the distal phalanx, articular fractures that involve
ligamentous attachments or tendon attachments of the phalanges, scaphoid polesmall fracture fragments or other small carpal fractures, and radial styloid fractures

Small Cannulated Ball Spike5294 [1.2 mm Hole]

Wire Hole for: 1,2 mm (.045") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

5294-01 [1.6 mm Hole]

Wire Hole for: 1,6 mm (.062") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

Designed by Benjamin C. Taylor, MD

Designed to help reduce a bone fragment and keep it reduced,
while the cannulation allows placement of a K-wire(up to 1.6 mm/.062") into the fragment
Helps to prevent slipping while inserting K-wires
Can serve as a handle for K-wire joysticks

PRODUCT NO:

8092

Overall Length: 4.5" (11,4 cm)
Handle Length: 3" (7,6 cm)
Ball Diameter: .275" (7 mm)

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

9

Holiday Self-Retaining
Carpal Tunnel Retractor
Designed by Allan Holiday, MD

PRODUCT NO:

1113

Overall Length: 6" (15,2 cm)

Burgess
Carpal Tunnel Retractor
Designed by Kraig Burgess, DO

Designed for exposure
during carpal tunnel surgery
PRODUCT NO:

1887

Overall Length: 4.25" (10,8 cm)
Blade Length: 12 mm
Blade Depth: 8 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Hagan Carpal Tunnel Release Sleeve
Designed by Hugh Hagan, MD

PRODUCT NO:

1150

Overall Length: 5" (12,7 cm)

Designed to protect the surrounding anatomy
while providing a sleeve within which to smoothly
advance a flat 4 mm beaver-style blade to divide
and release the transverse carpal ligament

Designed to use a Beaver-styleMini-Meniscus (Flat) 4 mm Blade.
Blade not included.

SleeveBlade Advancer with Blade(Blade not included)

Designed for use in a mini-open, non-endoscopic approach, thesleeve isolates the blade, providing protection to the surrounding
anatomy. The longer, bottom leading edge of the sleeve is inserted
between the median nerve and the transverse carpal ligament,
while the shorter, top leading edge provides lifting protection to
the structures above the ligament. The blade is then advanced
within the sleeve to complete the ligament release.

Evans Universal Carpal Tunnel Knife Guide

Designed to protect the median nerve while providing a choiceof grooved tracks for commercially available retrograde knives(that do not provide this feature) or for tenotomy scissors
Designed by Peter J. Evans, MD, PhD

Allows for smooth advance of the blade or scissors to divide the transverse carpal ligament.
Designed for a mini-open, non-endoscopic approach.

PRODUCT NO:

1128

Overall Length: 8" (20,3 cm)
Blade Guide Widths: 2 mm and 5 mm

10

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Ditmars Carpal Tunnel Release Set

Designed by Donald M. Ditmars Jr., MD

Designed to help retract and provide access for carpal tunnel release operations

PRODUCT NO’S:

1132-00 [Carpal Tunnel Release Set with Case]
Also Available Individually:

1132-01 [Large Curved Release Retractor]
Overall Length: 5" (12,8 cm)
Handle Length: 3" (7,6 cm)
Inside Tube Diameter: 7,5 mm

1132-02 [Small Curved Release Retractor]
Overall Length: 4.75" (12 cm)
Handle Length: 3" (7,6 cm)
Inside Tube Diameter: 4 mm

Small Curved
Release Retractor

Large Curved
Release Retractor

Small retractor for initial use in
carpal tunnel release operations

Retractor for carpal tunnel
release operations

4 mm inside tube diameter

7,5 mm inside tube diameter

1132-03 [Straight Carpal Tunnel Probe]
Overall Length: 7.5" (19,1 cm)
Handle Diameter: .25" (6,25 mm)

1025 [Sterilization Case]

Straight
Carpal Tunnel Probe
Probe to act as dilator to insert small
retractor for carpal tunner release operation

Lubahn Corkscrew

Prototype Shown

Designed by John D. Lubahn, MD

Prototype
Shown

Standard

Designed to help withremoval of tarsal and/or carpal bones
Aids trapezium removal during basal joint arthroplasty
when the bone is being removed as a unit
Can also be used to facilitate a proximal row carpectomy
as it fits the scaphoid, lunate, and triquetrum
May additionally be used to remove the pisiform in casesof arthritis of the piso-triquetral joint

Corkscrew Small Bone Manipulator

PRODUCT NO'S:

1191 [Standard]

Overall Length: 2.25" (5,7 cm)

1191-01 [Extended]

Overall Length: 6.5" (16,5 cm)

Extended

Designed by Raymond Wurapa, MD

Designed with an aggressive thread to aid in excising small bones of the hand and footThe quick-connect end allows the device to be inserted with ease under power with a standard drill attachment.
After insertion, the drill is detached and manual control over the process of extracting the bone can be
performed by hand, using either the disc on the shaft or attaching a handle.
Helps with removal of trapezium during basal joint arthroplasty
Helps with extraction of any carpal bones forwrist procedures: proximal row carpectomy (PRC),
partial wrist fusions, pisiform excision

PRODUCT NO’S:

1615

Overall Length: 4" (10,2 cm)
Length Beyond Disc: 2.25" (5,7 cm)
Length Beyond Line: .625" (1,6 cm)
Corkscrew Length: .375" (1 cm)
Optional:

S0113 [Universal Handle]
Overall Length: 4" (10,2 cm)

Shown with optional
universal handle
attached(Handle not included.)

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

11

Kakar Carpal Tunnel Retractors

Designed by Sanj Kakar, MD

Designed for maximum ergonomic positioning and soft
tissue retraction to permit release of the transverse
carpal ligament through a mini open technique

PRODUCT NO’S:

1126 [Small]

Overall Length: 6" (15,2 cm)
Blade: 15 mm Wide x 11 mm Tall

PRODUCT NO’S:

1159 [Standard Sharp Rake]
Overall Length: 4.5" (11,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1161 [Standard Blunt Rake]
Overall Length: 4.5" (11,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1162 [Standard Senn]

Chung T-Handle Retractors

1127 [Large]

Overall Length: 6" (15,2 cm)
Blade: 20 mm Wide x 15 mm Tall

Designed by Raymond Chung, MD

Designed with a T-handle for easier holding andto help reduce finger and thumb fatigue
Standard
Shaft Length

Overall Length: 4.5" (11,4 cm)
Blade Width: 6 mm
Blade Depth: 16 mm

Sharp Rake
Blunt Rake

1159-01 [Extended Sharp Rake]
Overall Length: 5.625" (14,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

Rake
Senn

1161-01 [Extended Blunt Rake]
Overall Length: 5.625" (14,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

Senn

1162-01 [Extended Senn]
Overall Length: 5.625" (14,4 cm)
Blade Width: 6 mm
Blade Depth: 16 mm

Hand/Finger Positioner

Extended
Shaft Length

Designed by Emad Aboujaoude, MS, MPAS, PA-C

Designed to help provide surgical positioning during
flouroscopy and fixation by isolating the operative
digit while retracting the unaffected digits
Radiolucent positioner can be steam or gas sterilized.
PRODUCT NO:

1134

Overall Length: 5.75" (14,6 cm)
Handle Width: 4.25" (10,8 cm)
Blade Width: 1.6" (4 cm)

12

Uses include but not limited to:
Intramedullary Metacarpal Screw
Phalanges CRPP
Digit Amputation
Digit Mass Excision
Finger Joint Fusion

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Swanson Elevator
Designed by Richard Ferkel, MD

Angular design helps to go around bone
for retraction and elevation — especially
useful in small bone surgery of thehand/wrist and foot/ankle
PRODUCT NO:

1644

Overall Length: 6.375" (16,2 cm)
Blade Depth: .75" (1,9 cm)

J.B. Redler Retractor
Designed by M.R. Redler, MD
PRODUCT NO:

1645

Overall Length: 5" (12,7 cm)

Uniquely balanced retractor for bone exposure
for a multitude of upper extremity procedures

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Double-angle design allows for ideal exposure with minimal effort to hold theretractor, while the assistant's hands are well out of the way of the exposure.
The aperture in the base of the handle allows the retractor to be attached via
a Penrose drain to the table for hands-free approach.

PRODUCT NO’S:

1665 [Blade: 6 mm Wide / 35 mm Drop]
Overall Length: 5.875" (14,9 cm)
Blade Width: 6 mm
Blade Drop: 35 mm

Modified Mini
Hohmann Retractors

1665-01 [Blade: 6 mm Wide / 17 mm Drop]

Used for small bone surgery

1666 [Blade: 8 mm Wide / 35 mm Drop]

Overall Length: 5.5" (14 cm)
Blade Width: 6 mm
Blade Drop: 17 mm

Designed by Jeffrey Lawton, MD

Overall Length: 5.875" (14,9 cm)
Blade Width: 8 mm
Blade Drop: 35 mm

1666-01 [Blade: 8 mm Wide / 17 mm Drop]
Overall Length: 5.5" (14 cm)
Blade Width: 8 mm
Blade Drop: 17 mm

OrthoLucent™
Mini Hohmann Retractors
Designed by Jeffrey Lawton, MD

Radiolucent, lightweight retractors
The carbon fiber PEEK material is strong,lightweight, completely radiolucent, can be
steam sterilized, and helps to prevent frommarring component surfaces.
PRODUCT NO’S:

1594-R [8 mm Blade]

Overall Length: 6.875" (17,5 cm)
Blade Width: 8 mm

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

1597-R [16 mm Blade]

Overall Length: 6.875" (17,5 cm)
Blade Width: 16 mm

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

13

Gap Clamp for Cortical Button Distal Bicep Repair
Designed by Corey Trease, MD

Designed to be used to help consistently set the gap
for the radius cortex between the distal biceps
stump and the cortical button

PRODUCT NO:

5262

Overall Length: 8" (20,4 cm)
Jaw Width Tapers from: 6 to 2.5 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Beard Distal Radius Wide Hohmann Retractor
Designed by David Beard, MD

Designed for distal radius and diaphyseal
fracture exposure, the wide blade design helps toprotect soft tissues, and the curved handle helps
provide improved access and visualization
PRODUCT NO:

5837-01

Overall Length: 5.375 (13,7 cm)
Blade Width: 1" (25 mm)

Lawton Distal Radius Mini Frame & Blade Set
Designed by Jeffrey Lawton, MD

Designed for self-retaining exposure for
distal radius and other small bone fractures

Short Blade Depth

Small & Large
Blade Depth

Set includes: (1) Frame,(2) Short Blades, (2) Small Blades.
Optional Large Blade available separately.

PRODUCT NO:

1578-00 [Set]
Set Includes / Available Individually:

1578-01 [Mini Frame]

Dimensions: 3" x 2.5" (7,6 x 6,4 cm)

1578-02 [Mini Short Blade](2) included in set, (1) with this product number
Overall Length: 2.5" (6,4 cm)
Blade Width: .625" (16 mm)
Blade Depth: .875" (22 mm)

1578-03 [Mini Small Blade](2) included in set, (1) with this product number
Overall Length: 2.625" (6,7 cm)
Blade Width: .625" (16 mm)
Blade Depth: 1.125" (29 mm)
Optional Blade / Not Included In Set:

1578-04 [Mini Large Blade]
Overall Length: 2.5" (6,4 cm)
Blade Width: .935" (24 mm)
Blade Depth: 1.125" (29 mm)

Short Blade Small Blade

14

WWW.INNOMED.NET   

Optional
Large Blade   FREE TRIAL ON MOST INSTRUMENTS

Chandran Distal Biceps Tissue Protector
Designed by Rama E. Chandran, MD

Designed to help protect tissue andexpose the radial tuberosity during
distal biceps tendon repair
Using downward pressure, theteeth help to engage bone tokeep the protector in place.

Helps to protect tissue and expose theradial tuberosity during distal biceps tendon repair

Also useful to help expose
the humerous during proximal
subpectoral biceps repair

PRODUCT NO:

3224

Overall Length: 8" (20,3 cm)
Handle Length: 5.5" (14 cm)
Blade Depth: 1.7" (4,3 cm)
Blade Width: .65" (1,7 cm)

Vaughan Distal Bicep
Tendon Repair Retractor
Designed by Roderick A. Vaughan, MD

Designed to retract in acontinuous way in three
directions, helping toprevent the surrounding vital
structures from entering
the field while drilling or
performing the repair work
PRODUCT NO:

3223

Profile View

Overall Length: 8.375" (21,3 cm)
Handle Length: 5.25" (13,3 cm)
Depth: 2" (5,1 cm)

Beard Distal
Bicep Retractor

PRODUCT NO’S:

5834-00 [Set – Retractor & Two Blades]
Available Individually:

5834-01 [Blade](1) blade with this product number, (2) Included in Set

Designed by David Beard, MD

Overall Length: 6.375" (16,2 cm)
Width: .625" (16 mm)

Designed to help
optimize surgical
exposure during anterior
single incision distal
biceps tendon reinsertion

5834-02 [Self-retaining Retractor]
Overall Length: 7.5" (19,1 cm)

The blade design features an anatomicallycontoured distal end to hug the radius cortex.
The smooth distal end helps to avoid deep
penetration, and the width matches the widthof the distal biceps tendon insertion site.
The narrow curved handle design helps to
optimize workspace and visualization.

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

15

Wurapa Swivel Blade Retractor
Designed by Raymond Wurapa, MD

Designed for forearm and wrist fracture exposure,the blades swivel for less stress on soft tissue
Swivel-blade technology helps to allow parallel deployment of retractor blades
to maximize wound exposure and minimize edge loading on surrounding
soft tissues. Parallel deployment of the retractor blades also helps prevent
rotation and migration of the retractor during a procedure.

PRODUCT NO’S:

1646-00 [Set]

Includes Retractor and Two Swivel Blades

Also available individually:

1646-01 [Retractor]

Overall Length: 5.125" (13 cm)
Opens to: 2.5" (6,4 cm)

1646-02 [Swivel Blade]

One blade with this product number, two included in set

Width: .9375" (24 mm)
Depth: .75" (19 mm)

Chung Weitlaner Retractors
Designed by Raymond Chung, MD

Longer prongs allow use ina small, but deep wound

Prong lengths of 25 mm and 30 mm
available with either sharp or blunt tips

3x4 Prongs

PRODUCT NO’S:
3x4 Prongs — Blunt Tips

3x4 Prongs — Sharp Tips

PRODUCT NO’S:
2x3 Prongs — Blunt Tips

2x3 Prongs — Sharp Tips

5065-01 [25 mm]

5066-01 [25 mm]

5065 [25 mm]

5066 [25 mm]

5067-01 [30 mm]

5068-01 [30 mm]

5067 [30 mm]

5068 [30 mm]

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)

25 mm

Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

30 mm

3x2 Prongs

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Williams Distal Radius
Fracture Retractor
Designed by Craig S. Williams, MD and Eric Dahlinger

Designed to provide excellent
exposure during fracture
reduction and plating
PRODUCT NO’S:

Left

1837-L [Left]

For Pins up to .045" (1.1 mm)
Overall Length: 4.5" (11,4 cm)
Blade Depth: 20 mm
Blade Width: 12.5 mm

1837-R [Right]

For Pins up to .045" (1.1 mm)
Overall Length: 4.5" (11,4 cm)
Blade Depth: 20 mm
Blade Width: 12.5 mm

16MADE FOR INNOMED IN

GERMANYRight

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Faillace Ambidextrous
Self-Retaining Retractor
Designed by John J. Faillace, MD

Handle can be rotatedaway from the surgeon after
insertion if desired
PRODUCT NO’S:

1580 [7 Teeth]

Overall Length: 7.5" (19,1 cm)
Prong Depth: 38 mm
Prong Width: 34 mm

1579 [4 Teeth]

Overall Length: 6" (15,2 cm)
Prong Depth: 38 mm
Prong Width: 18 mm

1579-01 [Small – 4x3 Teeth]
Overall Length: 5.25" (13,3 cm)
Prong Depth: 20 mm
Prong Width: 18 mm / 13 mm

Speed lock helps allow
precise control and prevent
unintended releaseWeinraub Joint and Calcaneal Spreader
Designed by Glenn M. Weinraub DPM, FACFAS

Designed to assist in the opening of small joints of the footand hand for the application of fusion and graft techniques
Provides excellent joint exposure without blocking intraarticular or osteotomy access.
Helps prevent slippage or falling out of the joint by placing the arms on eitherside of the area to be distracted, driving two pins and opening the joint.
PRODUCT NO’S:

1870 [Standard 1.6 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .062" (1/16") (1.6 mm)

ORIGINAL
DESIGN!

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY1872 [Standard 2.8 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .11" (7/64") (2.8 mm)

1870-SL [Speed Lock 1.6 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .062" (1/16") (1.6 mm)

1872-SL [Speed Lock 2.8 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .11" (7/64") (2.8 mm)

Gurbani Joint Distractor/Compressor
Designed by Naren G. Gurbani, MD

Pin Hole Sizes:.15" (3,5 mm) and.182" (4,5 mm)

Versatile joint distractor/compressor forarthroscopic or open procedures of foot,ankle, hand, and wrist joints
The surgeon puts the pins in the bone, then slides the holes ofthe device over the pins and distracts or compresses—the device
can be locked in either direction. Especially useful for arthroscopyof subtalar, talo-navicular, calcaneo-cuboid, and wrist joints. The
T-wrench helps provide precise, controlled manipulation.

PRODUCT NO’S:

4208-00 [Set]

Includes: Distractor/Compressor, T-Wrench, and Case

Available individually:4208-01 [Distractor/Compressor Only]

Dimensions: 6" w x 5" h (15,2 cm x 12,7 cm)
Distracts up to: 3" (7,6 cm) / Compresses down to: .5" (1,3 cm)

4208-TW [T-Wrench]

Dimensions: 3" w x 3" h (7,6 cm x 7,6 cm)

1025 [Sterilization Case]

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

17

Monaco Small Space Retractor
Designed modified by Spencer Monaco, DPM, FACFAS

Designed to retract adipose tissue
and surrounding soft tissue structures
through a small incision for open plantar
fasciotomies, neuroma excisions and thelateral release during bunion surgery
Also useful for various hand surgeries such as open carpal tunnel surgery.
PRODUCT NO:

1887-01

Overall Length: 4.25" (10,8 cm)
Blade Depth: 18 mm
Blade Width: 12 mm
Blade Lip: 3.5 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

WilsonTrigger Finger
Retractor
Designed by Ralph V. Wilson, MD
PRODUCT NO:

1884

Overall Length: 4.25" (10,8 cm)
Blades: 6.5 mm Wide x 10 mm Deep

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYHFD Self-Retaining
Small Bone Spreader

Versatile spreader featuring narrow tapered
blades which, when together, make a small wedge
to enter a tight bone interface or osteotomy
Blades feature a non-aggressive grip pattern thatcan be used when spreading apart bone as well as
providing retraction of soft tissue in a smaller wound.
PRODUCT NO:

1829

Overall Length: 4.5" (11,4 cm)
Blade Depth: 28 mm
Blade Width Tapers from: 8 mm to 5 mm

Hendren Neuroma Retractor
Designed by Douglas H. Hendren, MD

Narrow tines are delicate on tissue,
but sturdy enough to retract bone
Provides excellent exposure. Also helpfulin scaphoid fracture repair surgery.
PRODUCT NO’S:

1680-02 [Large]

Overall Length: 5.5" (14 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1680-01* [Small]

Overall Length: 4.25" (10,8 cm)

18

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Dodson Modular Retractor
Designed by Mark A. Dodson, MD

Allows the limb to be rotated (pronated or supinated)without loss of exposure. The hohmann retractors havethree hole sizes which allow for a variety of positioning
angle options using the teeth of the self-retaining
retractor, or can also be positioned in-between theteeth. The hohmann is placed around the bone,
and thus reduces the force on the soft tissues while
increasing exposure. Can be used in the forearmto treat radius and ulna shaft fractures, humerus
fractures, as well as in the leg for fibula fractures.

Designed to help expose a small to medium sizebone for internal fixation—can be used for distal
radius, ulna, humerus, and fibula fractures

PRODUCT NO’S:

Set consists of
one ratcheting selfretaining retractor,
1838-01 [Retractor Only]
two stainless steel
Overall Length: 5.5" (14 cm)
mini-hohmann
1838-02 [Blade Only – One] retractor blades, anda sterilization case.
Overall Length: 5.25" (13,3 cm)
1838-00 [Set]

Replacement Parts:

Blade Width: 3/8" (9 mm)

1025 [Sterilization Case Only]
Optional Parts — Not Included In Set:

1838-02R* [Radiolucent Blade Only – One]
Overall Length: 5.25" (13,3 cm)
Blade Width: 3/8" (9 mm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY* MADE EXCLUSIVELYFOR INNOMED IN

US Patent No. 9,161,745 B2

SWITZERLAND

Optional radiolucent carbon fiber PEEK composite blade
The optional radiolucent blade is made of a strong, lightweight
carbon fiber PEEK composite material, which is completely radiolucent,helps to prevent from marring component surfaces, and can be steam sterilized.

Calibrated Ortho Spreader
with Slotted Tips
Designed by Jason Bariteau, MD

A lamina spreader with a very thinclosed profile, designed to enabledistraction in tight spaces like thesubtalar and talonavicular joints
PRODUCT NO:

1841

Overall Length: 6.75" (17,1 cm)
Prong Length: .5" (12,7 mm)
Calibrations: 10 mm to 35 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Ortho Self-Retaining
Retractor with Pin Guides
Designed by Sean Dunn, DPM

Designed to distract a small
joint during fusion or osteotomyalignment surgery

1.800.548.2362       JULY 2023   

PRODUCT NO:

1842-02

Overall Length: 6.5" (16,5 cm)
Blade Width: 7 mm
Blade Extension (beyond guides): .4" (1 cm)
Blade Thickness: 1.68 mm
Pin Guide Length: 1.25" (3,2 cm)
Pin Guide Internal Diameter: .085" (2,1 mm)   HAND & WRIST INSTRUMENTS

19

HFD Compressor/Distractor

Dial mechanism helps allow precise control of inserted wires—for maintaining a position, compressing or distracting
A .125" (3,2 mm) pin can be used in the holes of the thumbwheel for leverage
Small: Two hole sizes allow for ease of pin size selection: .045" (1,1 mm) & .062" (1,6 mm)
Large: Two hole sizes allow for ease of pin size selection: .082" (2,0 mm) & .125" (3,2 mm)
Radiolucent arms are a steam sterilizable PEEK/Carbon Fiber composite

Radiolucent arms

PRODUCT NO’S:
SMALL

LARGE

1834 [Small – All Stainless Steel]1836 [Large – All Stainless Steel]

Dimensions: 51 mm x 57 mm
Maximum Arm Opening: 1.35" (3,4 cm)

Overall Length: 4" (10,2 cm)
Maximum Arm Opening: 2.25" (5,7 cm)

1834-R [Small With Radiolucent Arms] 1836-R [Large With Radiolucent Arms]

Radiolucent arms

Dimensions: 51 mm x 57 mm
Maximum Arm Opening: 1.35" (3,4 cm)

Overall Length: 4" (10,2 cm)
Maximum Arm Opening: 2.25" (5,7 cm)

Wurapa Small Joint Compressor and Distractor
Designed by Raymond K. Wurapa, MD

Designed to allow one-handed manipulation anddeployment once fixation pins are placed

Available with two holesizes on each instrument!

K-wires should be cut short above the pin guides
to allow full access to the operative site.
Compressor

Distractor

PRODUCT NO’S:DOUBLE HOLES:.045" (1,1 mm) & .062" (1,6 mm)

1751 [Compressor]

Compresses From: 28 mm
Overall Length: 4.625" (11,7 cm)

1752* [Distractor]

Distracts to: 46 mm
Overall Length: 4.625" (11,7 cm)

Designed to simplify several small joint procedures:
Preparation of small bone non-unions before bone grafting and fixation
Preparation of small joints for arthrodesis (e.g. partial wrist fusion)
Distract and better evaluate small joints before determining final management
Useful for intercarpal stabilization while performing ligament reconstructions(e.g. scapholunate ligament repair/reconstruction)

20

WWW.INNOMED.NET   SINGLE HOLE:.045" (1,1 mm) Hole

1753 [Compressor]

Compresses From: 28 mm
Overall Length: 4.5" (11,4 cm)

1754 [Distractor]

Distracts to: 46 mm
Overall Length: 4.5" (11,4 cm)

MADE EXCLUSIVELYFOR INNOMED IN*

GERMANY   FREE TRIAL ON MOST INSTRUMENTS

PRODUCT NO:

3654

Overall Length: 7.5" (19,1 cm)
Pin Hole Diameters: 4.5, 5.5, & 8.5 mm
Leg & Pin Hole Depth: .7" (17,5 mm)*Screwdrivers not included.

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Mantis Screwdriver Distractor
Designed by J. Albert Diaz, MD

Designed to help provide stable distraction acrossdifficult-to-reduce fractures using two seated screwdrivers*
Accommodates screwdrivers of varying sizefor use with both small and large fragment systems
Allows for distraction of difficult-to-reduce fractures
without the need to drill additional holes outside of the plate
The plate can be locked with a screw once length has been restored

Joint, Calcaneal andSmall Bone Distractors

Speed lock helps allow
precise control and prevents
unintended release

Two hole sizes and two arm
designs allow for easier pin size
selection and helps with distraction
in a variety of indications
Two hole sizes for ease of pin size selection:.062" (1,6 mm) & .094" (2,4 mm)
PRODUCT NO’S:
CLOSED ARMS

EXTRA SMALL
WITH SPEED LOCK4210-XSD [Extra Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 4.25" (10,8 cm)

EXTRA SMALL

CLOSED ARMS WITH SPEED LOCK

4216-XS [Extra Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 4.5" (11,4 cm)

Joint, Calcaneal, andSmall Bone Compressor

Designed for compression in
fracture and osteotomy procedures
Two hole sizes for ease of pin size selection:.062" (1,6 mm) & .094" (2,4 mm)

SMALL

PRODUCT NO’S:

4210-SC [Small]

Overall Length: 6" (15,2 cm)

4210-XSC [Extra Small]

Overall Length: 4.25" (10,8 cm)

EXTRA SMALL

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

21

Desai Curette Osteotomes

Cup Sizes:

Designed by Sarang Desai, DO

5 x 6 mm

The osteotome portion also can be used to"feather" the subchondral surface to expose
bleeding bone. It is also useful in instances ofobtaining autograft, as it can be used to create abone window and then remove cancellous bone.

8 x 10 mm

PRODUCT NO’S:

5241 [5 x 6 mm]

Overall Length: 8.25" (21 cm)
Osteotome Width: 3.5 mm
Osteotome Length: 3.5 mm from edge of cup

5242 [8 x 10 mm]

Overall Length: 8.25" (21 cm)
Osteotome Width: 6.5 mm
Osteotome Length: 3 mm from edge of cup

Designed to remove bone andcartilage, helpful for preparing joint surfaces
for fusion, allowing easy removal of osteophytes
and cartilage without having to switch instruments

Hemisphere Curettes

Designed by Richard Wittock, DPM and Rob Baglio, DPM

Designed for small joint surgery
5 mm

PRODUCT NO’S:

5345

Overall Length: 5.75" (14,6 cm)
Curette Diameter: 5 mm

5349

Overall Length: 5.75" (14,6 cm)
Curette Diameter: 9 mm

9 mm

Ring Curettes

PRODUCT NO’S:
Straight Shaft

PRODUCT NO’S:
Bent Shaft

MADE FOR INNOMED IN

Overall Length: 8.75" (22,2 cm)

GERMANY

5150 [3 mm, Straight] 5152 [6 mm, Straight]
Ring Diameter: 3 mm

Ring Diameter: 6 mm

5154 [8 mm, Straight]
Ring Diameter: 8 mm

MADE FOR INNOMED IN

Overall Length: 8.625" (21,9 cm)

5156 [3 mm, Bent]

Ring Diameter: 3 mm

Micro Curettes

GERMANY

5157 [6 mm, Bent]

Ring Diameter: 6 mm

5158 [8 mm, Bent]

Ring Diameter: 8 mm

PRODUCT NO’S:
Straight Micro Curettes

Four cup sizes,
straight or 45°
angled-end shaft

Overall Length: 9.75" (24,8 cm)
Shaft Length: 4.5" (11,4 cm)
4242 Cup Size 24240 Cup Size 14244 Cup Size 4/04246 Cup Size 6/0
Angled Micro Curettes

Overall Length: 9.75" (24,8 cm)
Shaft Length: 4.5" (11,4 cm)
4242-01 Cup Size 24240-01 Cup Size 14244-01 Cup Size 4/0
4246-01 Cup Size 6/0

22

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

PRODUCT NO’S:
Offset Chisels

Ortho Mini Chisels

Straight Chisels

1169-1 [1 mm Offset Chisel] 1170-3 [3 mm Straight Chisel]
Overall Length: 6.25" (15,9 cm)
Chisel Width: 1 mm

Overall Length: 6.4" (16,3 cm)
Chisel Width: 3 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 2 mm

Overall Length: 6.4" (16,3 cm)
Gouge Width: 4 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 3 mm

Overall Length: 6.4" (16,3 cm)
Gouge Width: 5 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 4 mm

MADE EXCLUSIVELYFOR INNOMED IN

Mini orthopedic chisels, straight and offset,with straight and ergonomic handles

1169-2 [2 mm Offset Chisel] 1170-4 [4 mm Straight Chisel]1169-3 [3 mm Offset Chisel] 1170-5 [5 mm Straight Chisel]1169-4 [4 mm Offset Chisel]1169-5 [5 mm Offset Chisel]

5 mm Offset Chisel ShownGERMANY

Overall Length: 6.25" (15,9 cm)
Chisel Width: 5 mm

4 mm Straight Chisel Shown

Ortho Mini Gouges

Mini orthopedic gouges with ergonomic handles, designed forbone resection in small areas and resection of periosteum

PRODUCT NO’S:

1168-2 [2 mm Gouge]

Overall Length: 5.75" (14,6 cm)
Gouge Width: 2 mm

Small, thin osteotomes helpful in osteophyte
and cement removal in total joint surgery.
Larger handle helps with better control.

GERMANY

1168-3 [3 mm Gouge]

1168-6 [6 mm Gouge]

1168-4 [4 mm Gouge]

1168-7 [7 mm Gouge]

1168-5 [5 mm Gouge]

1168-8 [8 mm Gouge]

Overall Length: 5.75" (14,6 cm)
Gouge Width: 3 mm
Overall Length: 5.75" (14,6 cm)
Gouge Width: 4 mm

5 mm Gouge Shown

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 5.75” (14,6 cm)
Gouge Width: 5 mm

Overall Length: 5.75” (14,6 cm)
Gouge Width: 6 mm
Overall Length: 5.75" (14,6 cm)
Gouge Width: 7 mm
Overall Length: 5.75" (14,6 cm)
Gouge Width: 8 mm

Mini-lexer Osteotomes

Helpful in osteophyte and cement removal
PRODUCT NO’S:

5270-01

5270-035270-025270-04

Blade Width: 4 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)
Blade Width: 6 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

Blade Width: 10 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)
Blade Width: 12 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

MADE FOR INNOMED IN

GERMANY

Hooked Bone Awls
Designed by Reza Firoozabadi, MD“Shoulder hook” awls designed
to help with manipulation ofbone fragments for fixation
PRODUCT NO’S:

5078 [Standard]

Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)

5078-01 [Long]

Overall Length: 13.375" (34 cm)
Handle Length: 6" (15,2 cm)

1.800.548.2362       JULY 2023   

Standard hooked bone awl being used to gain length
to assist with reduction of a fibula fracture. A 2mm
pilot hole is made to seat the tip of the bone awl.

   HAND & WRIST INSTRUMENTS

23

Fracture Reduction Pick

Used to align bone fragments,
and to pick away tissue andbone fragments
PRODUCT NO:

S0129

Overall Length: 6.25" (15,9 cm)

Flexible Osteotome Instruments

Choice ofHandle Style

An assortment of flexible osteotome blades
useful in foot & ankle surgery procedures

Handle with Quick-Coupling End

Handle with Locking Nut

3" Osteotome Blades

Sharp, flexible blades are well suited for loosening implants from
cement or bony ingrowth fixation
Various blade widths and profiles allow great flexibility to follow the
implant contours
Modular handle is made of high impact surgical stainless steel andhas a quick-coupling positive locking mechanism for ease of use andquick blade changes
Slap hammer threads into the handle and is designed to facilitate
blade removal
Optional Strike Plate can be attached to the Handle for direct striking
with a mallet
Optional Curved Chisel Blades can be used to help loosen the cement/
prosthesis interval in total ankle revisions. The curved design is useful
in working around pegs & fins to get posterior cement access. Also
helpful with removal of other implants, i.e. shoulder, knee, femoral, etc.

2.5" Chisel Blades

PRODUCT NO’S:
Individual Instruments Available Separately

S1002 [Osteotome Blade] 3" (7,6 cm) x 8 mm
S1003 [Osteotome Blade] 3" (7,6 cm) x 10 mm
S1004 [Osteotome Blade] 3" (7,6 cm) x 12 mm
S1005 [Osteotome Blade] 3" (7,6 cm) x 20 mm
S1006 [Curved Osteotome Blade] 3" (7,6 cm) x 12 mm
S1020 [Handle with Quick-Coupling End] 6" (15,2 cm)
S1021 [Handle with Locking Nut] 6" (15,2 cm)
S1020-SP [Strike Plate for Handle] Diameter 1.625" (4,1 cm)
S1222 [Chisel Blade] 2.5" (6,4 cm) x 8 mm
S1223 [Chisel Blade] 2.5" (6,4 cm) x 10 mm
S1224 [Chisel Blade] 2.5" (6,4 cm) x 12 mm
S1225 [Chisel Blade] 2.5" (6,4 cm) x 20 mm
S1228 [Chisel Blade] 5" (12,7 cm) x 10 mm
S1233-L [Left Curved Chisel Blade] 1.5" (3,8 cm) x 8 mm
S1233-R [Right Curved Chisel Blade] 1.5" (3,8 cm) x 8 mm
S2007 [Slap Hammer] 12" (30,5 cm)

Curved chisel
design allows
working aroundcomponent
pegs, fins, etc.

1.5" Curved Chisel Blades

Strike Plate for Handle

Slap Hammer

Complete Set with more
options available onlineat www.innomed.net

Medial and Lateral Curve Radial Blades designed by Henry Boucher, MD
Curved Chisel Blades designed by William McMaster, MD

Narrow Cement Removal Gouge, Short

Bevel side awayfrom component

Mueller-Type Cement Removal Instruments
Useful for cement removal in the ankle
Also helpful in hip, knee, and shoulder surgery.
PRODUCT NO’S:
Individual Instruments Available Separately

Offset Chisel

S7505 [Narrow Cement Removal Gouge, Short]
Shaft Length: 10 cm
Gouge: 9 mm, negative

S7520 [Offset Chisel]
Cement Removal Osteotome, Short

S7595 [Cement Removal Osteotome, Short]
Shaft Length: 15 cm
Osteotome: 8 mm

4.4 mm Drill & Drill Guide

S7540 [4.4 mm Drill]
S7545 [4.4 mm Drill Guide]
S7570 [Cross Bar]
Complete Set with more options
available online at www.innomed.net

Cross Bar24

Shaft Length: 15 cm
Chisel: 9 mm

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Extended Drill Sleeves

Designed by Reza Firoozabadi, MD

Designed to help reduce fractures when K-wires are passed
through, the extra long drill sleeve helps to protect soft tissues
and prevent the need for stacking two drill sleeves
2.4/1.8 mm
2.7/2.0 mm
3.5/2.5 mm

Serrated tips allow for better grip whendrilling at an angle or when pushing afracture fragment to assist with fracture
reduction
Sleeve can be used as a reduction aid withplacement of a kirschner wire through sleeve
Collaborated tips which allow placementof appropriate size drills for lagging bytechnique — as an example a 2.5 end
will fit into a 3.5 drill hole

Case example ofusing modified
3.5/2.5 mm drill
sleeve for placing
3.5 mm screws in
a forearm fracturecase. Note how
extended sleeve
protects soft tissues
during drilling.

PRODUCT NO’S:

3014-00 [Set of Three]

ProfileSet Includes/ Available Separately:

3014-01 [2.4/1.8 mm]

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

Extra Long Guide Sleeve Length: 2.25" (5,7 cm)

Larger Diameter

Serrated Sleeve Ends

3014-02 [2.7/2.0 mm]

Collaborated tip end diameters
allow placement of appropriate size drills
for lagging by technique — as an example a
2.5 end will fit into a 3.5 drill hole.

Smaller Diameter

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

3014-03 [3.5/2.5 mm]
Serrated Sleeve Ends

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

Roberts Pin Bending Cannula Set
Designed by David Roberts, MD

Helpful when removing a pin
which has been cut flush to thebone which can be hard to grasp
with standard tools.
PRODUCT NO’S:

2113-00 [Set of 3 Sizes]
Set Includes / Available Individually:

2113-01 [2 mm]

Designed to help bend the end of a flexible
intramedullary pin, which has been cut flush to thebone, for better grasping during pin removal
After exposing the pin end, the cannula helps bend
the pin for better access for the removal
instrument while maintaininga small incision.

For 1.5 - 2.0 mm flexible nails
Overall Length: 5.5" (14 cm)

2113-02 [3 mm]

For 2.5 - 3.0 mm flexible nails
Overall Length: 5.5" (14 cm)

Note 2.0 mm end ofdrill sleeve placed
into a predrilled 2.7
mm hole, utilized asa lag by technique
2.7 mm screw.

Features
Beveled edge that allows
easy capture of end of pin
Cannula can be used as a
trephine for pins with bonyovergrowth
Reusable - thick walls
withstand repeated uses

4 mm
3 mm
2 mm

2113-03 [4 mm]

For 3.5 - 4.0 mm flexible nails
Overall Length: 5.5" (14 cm)

Technique

Distal Radius – 1.5 cm Incision

Proximal Ulna – 1.5 cm Incision

Advance cannula over the exposed pin.
The beveled end makes this easier andcan be used as a trephine in when bonyovergrowth is present.
Slowly use the tool to bend theend of the pin away from bone, whichhelps with pin removal.

The bent end can be removed
with standard pliers.

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

25

K-Wire Bender/Cutter
1 and
1.2 mm

1.2 and
1.6 mmThe right slot of the
instrument’s lower
jaw can hold K-Wireswith a diameter of
1.2 mm or 1.6 mm.
The smaller left slot
can hold K-Wires
measuring 1 mm or
1.2 mm in diameter.

Bending

Smooth Bending
Clean Cutting

Designed to bend a K-Wire while extending
from bone without applying mechanical strain

The K-Wire only needs to extend 20 mmfrom the skin surface to be bent.
Can bend and cut K-Wires
measuring 1 to 1.6 mm(.039–.062") in diameter
PRODUCT NO:

2111

Overall Length: 6.5" (16,5 cm)

With the jaw of the instrument
opened wide, the K-Wire isinserted from the side into oneof the slots of thelower jaw. During bending, theK-Wire is forced backwards by thenose of the upper jaw and guided
by a small groove.

Cutting

The K-Wire is inserted into
the cutting groove and thebender/cutter cuts by shearing(like a cigar cutter), notcrushing. The result is a clean
and burr-free cut surface.

Stanton Bent Pin Removal Pliers
Designed by John Stanton, MD, FACS
PRODUCT NO:

1894

Overall Length: 6.5" (16,5 cm)
Jaw Length: 1.65" (4,2 cm)
Instrument Width: 1 cm

Pin Puller - Small

Small size allows for use
in a small incision to help
with removal of a 2 mm or
smaller k-wire pin
PRODUCT NO:

3033

Overall Length: 6.5" (16,5 cm)
Jaw Width: 6,2 mm tapering to 3 mm at end
Jaw Height: 11,7 mm

Fromm Triangles

Designed by S.E. Fromm, MD
Extra Small Triangle designed by S.E. Fromm, MD & Kenneth Merriman, MD

Radiolucent triangles are useful for wrist
arthroscopy and allow for intraoperative fluoroscopy
Helps support the wrist and forearm during wrist arthroscopy procedures,
while allowing for traction on the opposite side. Sterilizable triangle canbe covered with a sterile towel for the procedure.
PRODUCT NO’S:

2760-01 [11"] Base: 6" (15,2 cm), Height: 11" (27,9 cm)
2760-XS [8.5"] Base 5" (12,7 cm), Height: 8.5" (21,6 cm)

26

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Wixted Fracture Distractor
Designed by John J. Wixted, MD

Designed to provide opposing leverage to help bring
the fibula (or other bone) back out to its proper length
after it has been shortened by a fracture

A 3.5 mm screw is temporarily placed
above a plate, providing a source ofleverage for the screw holding end of
the distractor. The curved peg-shaped
tip is then placed into a hole in thebone plate, and the distractor isactivated to bring the bone back to its
proper length before fixation.
PRODUCT NO:

1882

Curved Peg-shaped Tip
Fits securely into a hole ina bone plate for leverage

Overall Length: 7" (17,8 cm)
Cut-out for Screw
Provides a secure source ofleverage against a temporarily
placed 3.5 mm screw

PRODUCT NO’S:

7653-00 [Set w/Case]
Individual Parts:

7653-01 [1.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

7653-02 [2.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

7653-03 [3.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

Lawton Screw Extractors

Designed by Jeffrey Lawton, MD

Designed to help extract mini and micro
fragment screws; small cannulated
screws; or headless screws

1025 [Sterilization Case]

Lawton Broken Screw Extractor
Designed by Jeffrey Lawton, MD

Designed to help remove broken or
stripped screws (1 mm-2 mm)
PRODUCT NO:

7653-04

Overall Length: 4" (10,2 cm)
Handle Width: 3" (7,6 cm)

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

27

Wilke Angled Blunt Nose Scissors
Designed by Benjamin K. Wilke, MD

Allows blunt dissecting around criticalstructures (nerves, vessels, etc.)
while maintaining a cutting surface
for fascia. The tool's blunt ends canalso be used for cauterizing andgrabbing small vessels.
PRODUCT NO:

3078

Overall Length: 6" (15,2 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Scissors

Orthopedic Needle
Holder/Scissors

Holding
Tips

Drive a needle and cut a suture
without changing instruments
PRODUCT NO’S:
Standard Tips

MADE FOR INNOMED IN

GERMANY

3070 7.0" (17,8 cm)
Tungsten Carbide Tips

3055 5.5" (14 cm)
3065 6.5" (16,5 cm)
3075 7.0" (17,8 cm)

Longer sizes arehelpful in orthopedics

Rogozinski Locking Needle Driver/Scissors

Designed with a quick lock & release handle, can drivea needle and cut a suture without changing instruments
Scissors
Holding
Tips

PRODUCT NO’S:

3083 [Standard]

Designed by
Chaim Rogozinski, MD

3084 [Large]

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 6.5" (16,5 cm)

Overall Length: 7.75" (19,7 cm)

GERMANY

Sweed Dissecting Scissors
Designed by Tamer Sweed, FRCS (Orth)

Designed with a blunt, flat bar
fixed to the lower limb, the scissors
also act as a dissector to protect
underlying vital structures
PRODUCT NO:

3081

Overall Length: 6.625" (16,8 cm)
Bottom Pad: 16 mm x 6 mm)
Pad Extension Beyond Scissor: 6 mm

28

WWW.INNOMED.NET   

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   FREE TRIAL ON MOST INSTRUMENTS

Mazzara Rongeur for Small Bones
Designed by James T. Mazzara, MD

Designed for bone and soft tissue removal
in small joint surgery, the pistol grip handle
lessens hand fatigue and slippage, andallows for better visualization

2 x 10 mm
Jaw Bite4 x 10 mm
Jaw Bite

PRODUCT NO’S:

1765-04

Jaw Bite: 2 x 10 mm
Overall Length: 9" (22,9 cm)

1765-05

Jaw Bite: 4 x 10 mm
Overall Length: 9" (22,9 cm)

2 x 10 mm

4 x 10 mm

Two Jaw Sizes Available

Yezerski Small Bone Rongeurs
Designed by John Yezerski, MD

Designed for small bone
applications in the hand and footPRODUCT NO’S:

Small

1789 [Small]

Overall Length: 7.125" (18,1 cm)
Jaw Width: 4 mm
Jaw Bite Width: 3 mm
Jaw Bite Length: 20 mm

1789-01 [Extra Small]

Extra Small

Overall Length: 4.5" (11,4 cm)
Jaw Width: Tapers from 4,6 mm to 2 mm
Jaw Bite Length: 11 mm

Bates Needle Holder with Suture Cutter

No switching between needle
driver and scissors, or need forassistant to cut sutures for you
Cutting with opening of forceps
reduces possibility of damage
to surrounding tissues
Sliding the instrument down to
the suture knot allows quick andconsistent 2 mm suture tails
Slip the suture strands into
the suture cutting slot andslide the closed instrument
along until desired length oftail is achieved, then open the
instrument to cut the sutures

Designed by James E. Bates, MD

By trapping the suture and cutting when the forcep
is opened, helps to reduce stresson the surgeon’s hand

PRODUCT NO:

3071

Overall Length: 8.125" (20,6 cm)
Jaw Width: .25" (6,4 mm)
Open Jaw Length: .5" (12,8 mm)

Suture Cutter
Needle Holder

Stanton Needle Driver

Designed by John L. Stanton, MD, FACS

Allows a heavy cutting needle such as anOS-6 to be pushed through cancellous bonewhen re-attaching muscle or tendon
The groove captures the outer (convex) side of the needle andprevents the needle from spinning even when applying significant
pressure. Useful for reattaching the rotator cuff in rotator cuff
repairs, as well as in attaching suture anchors.
PRODUCT NO:

3042

Overall Length: 6.75 (17,1 cm)
Jaw Width: .25" (6,3 mm)

1.800.548.2362       JULY 2023      HAND & WRIST INSTRUMENTS

29

Universal Bone Grafting/Impacting Forceps

Designed by J. A. Amis, MD

Bone graft can be grasped, placed & impacted
without changing hands or instruments

1/8"
3/16"
1/4"
5/16"
(3,2 mm) (4,8 mm) (6,3 mm)(8 mm)
Diameter ends at actual size (closed forceps)

The forceps are designed with grasping ends for delivery
of bone graft. When the graft is in place, the forceps are closed,
which forms the ends into an impacting punch. A striking platform
is attached to the end of the forceps for tapping and tamping the graft.
Four end diameters are available in two lengths.
PRODUCT NO’S:
Short: 6" (15,2 cm) Length

5010-01 1/8" (3,2 mm) Diameter End
5010-02 3/16" (4,8 mm) Diameter End
5010-03 1/4" (6,3 mm) Diameter End
5010-04 5/16" (8 mm) Diameter End

Long: 10" (25,4 cm) Length

5050-01 1/8" (3,2 mm) Diameter End
5050-02 3/16" (4,8 mm) Diameter End
5050-03 1/4" (6,3 mm) Diameter End
5050-04 5/16" (8 mm) Diameter End

When the forceps areclosed, they form into
an impacting punch

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Faillace Bone Impact/Graft Forceps
Design modification by John J. Faillace, MD, FAAOSLong vertical grooves at the tip are designed
to deliver graft into a small space, where afreer elevator can be used to push the graftdown into the space, then the closed flat end
can be used to tamp down the graft
PRODUCT NO:

5011

Overall Length: 5" (12,7 cm)
Tip Diameter When Closed: 3,2 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Vaughan Endzone Retractor
Designed by Roderick Vaughan, MD

Designed for use when placing
the end screws while platinga fracture using a minimallyinvasive technique
PRODUCT NO:

1766

Overall Length: 8.75" (22,2 cm)
Deep Depth: 45 mm
Deep Internal Width: 14 mm
Shallow Depth: 25 mm
Shallow Internal Width: 12 mm

BeforeAfter

The “U”-shaped wall design
helps allow the maximal exposure
along the length, or "endzone",of an incision while maintaining
adequate width and retraction along
the sides of the exposure.

Syringe not included.

Gray Syringe Assist with Ergonomic Handle
Designed by Robert Gray, MD

For use in the O.R or the office, the design helps
to prevent hand fatigue and pain when injectingwith a 20mL syringe over multiple cases

30

PRODUCT NO:

8988

Overall Length – Closed: 5.25" (13,3 cm)
Overall Length – Open: 7.5" (19,1 cm)
Height: 5" (12,7 cm)
Syringe Diameter: 21 mm

WWW.INNOMED.NET   Patent Pending   FREE TRIAL ON MOST INSTRUMENTS

Sarraf TiN Coated
Cement Removal Forceps
Designed by Khaled M. Sarraf, MD

Ultra hard titanium nitride coating helps to extend
forceps life by increasing surface hardness,prolonging sharpness, and resisting chemicals
and corrosion, while helping to eliminate metaltransfer and protect the implant surface.
PRODUCT NO’S:

5039 [Straight]

Overall Length: 6" (15,2 cm)

5041 [Angled]

Overall Length: 6.125” (15,6 cm)

Adson Forceps with Cobb Elevator End
Designed by Oscar Castro-Aragon, MD

Has the advantages of having a Cobb
tip at the end of an Adson forceps
Allows the opportunity to do soft tissue dissection, cleaning of thebone or bone fragments in a fracture, push bone fragments to holda reduction in a fracture, separate soft tissue, and turn it around topick up tissue without having to switch instruments back and forth.

The curved semicircular tip is congruent
to most tibial plates and femoral condylar
implants, helping to facilitate removal of excess
cement, especially at the tight posterior aspect
The small scoop-end tip assists in excising
unset cement
Ultra hard titanium nitride coating helps
to extend curette life by increasing surface
hardness, prolonging sharpness, and resistingchemicals and corrosion, while helping toeliminate metal transfer and protect the
implant surface

PRODUCT NO:

1166

Overall Length: 4.75" (12,1 cm)
Tip Width: 2.4 mm (2,4 mm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Sarraf Cement Trimmer
Designed by Khaled M. Sarraf, MD

PRODUCT NO:

Two-in-one instrument designed for cement
removal during arthroplasty surgery

Bozeman Cement Trimmer

5212

Overall Length: 7.75" (19,7 cm)

Combines the two most common cement trimming tools into one

Designed by Daniel M. Gannon, MD

The tool has a blunt blade tip on one end to help with separationof the trimmed cement. The angled curette end helps gather thetrimmed cement. The thin shank and angled curette can reach into
tight spaces such as the back of the implants to remove excess
cement. The ends are titanium nitrite coated to help eliminate
metal transfer.
PRODUCT NO:
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

5245

Overall Length: 8.5" (21,6 cm)

Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:

31

PRSRT. STD.
U.S. POSTAGE
PAID
GRAND RAPIDS, MI

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 748

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

Chandran Distal Biceps
Tissue Protector
Designed by Rama E. Chandran, MD

Designed to help protect tissue andexpose the radial tuberosity during
distal biceps tendon repair
Using downward pressure, the teeth
help to engage bone to keep theprotector in place.

Helps to protect tissue and expose theradial tuberosity during distal biceps tendon repair

Also useful to help expose the humerous
during proximal subpectoral biceps repair

Also useful to help expose the humerous
during proximal subpectoral biceps repair.
PRODUCT NO:

3224

Overall Length: 8" (20,3 cm)
Handle Length: 5.5" (14 cm)
Blade Depth: 1.7" (4,3 cm)
Blade Width: .65" (1,7 cm)

INNOMED, INC

Innomed-Europe LLC

TOLL FREE 1.800.548.2362

Tel 912.236.0000
Fax 912.236.7766

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74

Innomed-Europe GmbHISO 13485:2016

103 Estus Drive
Savannah, GA 31404

Villingen-Schwenningen,Deutschland
Tel 0049 (0) 7720 46110 60

innomed.net
info@innomed.net© 2023 Innomed, Inc.
All Rights Reserved

07
23

www.innomed-europe.com
orders@innomed-europe.com

INNOMED.NET

info@innomed.net