PressureSensors
CONTENTS
Application and Caution

2SD Series/ Semiconductor Pressure Sensor Chip

3SP Series/ Semiconductor Pressure Sensor

4SM Series/ Semiconductor Pressure Sensor

5

SSP Series/ Semiconductor Pressure Sensor

6

SPAN Series/ Pressure Sensor Non-influenced by Noise

7

MKT-TAISEI Co.,Ltd.

Pressure Sensor Application
Micro-Pressure Range

Gas pressure control, washing machine water level control, filter clogging detection.
FA equipment such as vacuum cleaner clogging detectors, electronic blood pressure meter,

Low-Pressure Range

weather forecast equipment, altimeter, water pressure detector, water depth measurement
device, vehicle engine controller, compressor pressure controller, vacuum absorber and noncontact switches.

Low-Pressure Range

Vehicle suspension, vehicle break controller, air conditioner refrigerant pressure controllers
and hydraulic controllers for building and construction equipment.

CAUTION

1) Use within the specified conditions. No warranty is made over specified conditions of use.
2) Avoid measuring corrosive pressure medium or in catalysis, unless otherwise, it would be thecause of defects.
3) Avoid measuring pressure medium in high temperature, unless otherwise, it would be the causeof defects.
4) Avoid measuring flammable pressure medium, unless otherwise, it would be the cause ofexplosion or fire.
5) Avoid disassembling, taking off a part or changing it.

Semiconductor Pressure Sensor Chip / SD Series
FEATURES
Corresponds to customization from design to mass production.
Small chip size. ( 1 mm ~ 3 mm )Two model types are available. ( absolute pressure and gauge pressure )Wide pressure specification. ( 10k ~ 50M Pa or 0.1 ~ 500 kg/cm2 )Low cost, high accuracy, high reliability, high breaking pressure resistance.

MODELSD 20 0C A 103
Rated Pressure
Type of Pressure A: Absolute Pressure
G: Gauge Pressure
Chip Size

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT

SD200C

SD250N

5463
45

1.25

23

1.25

0.75

Pressure

1261654

123

3.1

2.6

2.1

1.1

SD300C

Pressure

Pressure

162

53

4

1.25

SD101K

Unit : mmPressure

0.8

0.8
Output
5Input2

Input43

Input1

Output
16

Output326
Output SubstrateOutput
15InputInput26Input
34SubstrateInput2Output
16Output354
Output Substrate5Input

4Substrate

SPECIFICATIONS
SD101K-A

Model No.
Rated Pressure[ Pa ]

Pressure Type

SD200C-A or G

SD300C-G

SD250N-A
103

303

503

104

504

101

201

501

1M 3.5M 100k 200k 500k 1M 1.5M 1M
Absolute or Gauge
Absolute

3M

5M

10M 20M 35M 50M

10k20k50k

103

353

102

202

502

103

Maximum Overload Pressure

153

204354Absolute

Gauge

Rated Pressure X 3

Temperature Range[ C ]

20 ~ 100

20 ~ 10020 ~ 100

Excitation Current[ mA ]

1.0

1.0

1.0

Bridge Resistance[ k ]

41514.51

Offset Voltage[ mV ]

025

025025

Output Span Voltage[ mV ]

60 ~ 140

60 ~ 140

50 ~ 80

Linearity[ %FS ]

0.3

20 ~ 1001.5

1.051030

025

60 ~ 140

0.3

0.3

Offset Voltage Temperature Characteristics * 1 [ %FS/C ]

0.1

0.1

0.1

0.5

0.1

Span Voltage Temperature Characteristics * 1[ %FS/C ]

0.05

0.05

0.05

0.2

0.05

Chip Size[ mm ]1.10.2
0.1

2.10.20.1

2.6 0.20.1

3.1 0.2
0.1

Chip Height[ mm ]

0.750.2

1.25

1.25 0.2

1.25 0.2

Stand Hole Diameter (Gauge Pressure)[ mm ]



0.8 0.1

0.2

0.80.1

0.5

NOTE *1 The temperature range is 0 to 60C. If there is no specification, the temperature range is 255C.
These specifications are examples. Please contact us for other specifications.

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

0.3

1.5e-sd-p01-11

Semiconductor Pressure Sensor / SP Series
FEATURESThe SP series is an inexpensive plastic package type ( DIP type ) .
These units correspond to regular equipment,
measurement equipment, home electronic equipment,
medical equipment and others.

MODELSP2 0 C G 102 U
Terminal Direction
Rated Pressure
Type of Pressure G : Gauge

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENTSP20CG

124812

0.8

4

0.88

6 5 4
1

15.24

R

13.5

13.5

15.24

R

1

30.25

36 5 4U

0.25

SP20CG

Unit : mm

50.

C

15x.

6.8ma

5

5
Input2.54

Input2152.542.54x.

Output
16

Output
3ma

Input
20.



1 2 3
2.54C

1 2 3

Output
16Output
34Substrate5Input

4Substrate

SPECIFICATIONSSP20CGModel No.
Rated Pressure[ kPa ]

101201501102202

1020

50

100

100 ~ 200

Pressure Type
Temperature range

20 ~ 100
Non-Corrosive Gases[ mA ]

1.5

1.0

Maximum Overload Pressure

Rated Pressure X 3

Output Span Voltage[ mV ]

Brige Resistance[k]Offset Voltage[ mV ]

X 1.5

60 ~ 14051025020

Offset Voltage Temperature Characteristics *1 [ %FS/C ]

0.5

0.1

Span Voltage Temperature Characteristics *1 [ %FS/C]

0.2

0.05

1.50.3

Linearity

103

Gauge
[ C ]

Pressure Medium
Excitation Current502

100 ~ 500 100 ~ 1000

[ %FS ]NOTE *1 The temperature range is 0 to 60C. If there is no specification, the temperature range is 255C.
These specifications are examples. Please contact us for other specifications.

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-sp-p01-12

Semiconductor Pressure Sensor / SM Series
FEATURESThe SM series are high precision and high durability metal
can package type ( TO-5 CAN equivalent package ) .

MODEL

SM20C G 102
Rated Pressure
Type of Pressure A : Absolute Pressure
G : Gauge Pressure

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT
137.9

Unit : mm

5.610SM10C313SM20C

0.45

7.9

465
Bottom View

Input5
1 6
Output

5.6

15
10

12

Output
Substrate343
Input23Low Pressure High Pressure
7.91510SM30C

465
Bottom View

1 6
Output5.6
12

Output
Substrate
343

2.5

4
65
Bottom View

Input
5

0.45

8.4

Input245

10.6

Input5

2.545

0.45

8.4

3



High Pressure

10.6

Output
Substrate34
Input
245

10.6
8.4

12

High Pressure

1 6
Output

SPECIFICATIONS
SM10C

Model No.
Rated Pressure[ kPa ]

501102202

102050

100

100 ~ 200502

20 ~ 100
Non-Corrosive Gases[ mA ]

1.5

1.0Maximum Overload Pressure

Rated Pressure X 3

Output Span Voltage[ mV ]

Bridge Resistance[k]Offset Voltage[ mV ]

60 ~ 14051025020

Offset Voltage Temperature Characteristics *1 [ %FS/C ]

0.5

0.1

Span Voltage Temperature Characteristics *1 [ %FS/C ]

0.2

0.05

[ %FS ]
1.5Linearity
NOTE *1 The temperature range is 0 to 60C. If there is no specification, the temperature range is 255C.
These specifications are examples. Please contact us for other specifications.

0.3

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/103

100 ~ 500 100 ~ 1000

Absolute or Gauge[ C ]

Pressure Medium
Excitation Current

SM30CA or G

201

Pressure Type
Temperature Range

SM20C

101

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-sm-p01-13

Semiconductor Pressure Sensor / SSP Series
FEATURESThe SSP11 series is a small plastic package type.
Built-in amplification and temperature compensation circuits,eliminate the need for circuit design and characteristic adjustment.
Terminal types are DIP and SMD.
It can be used for various devices such as general-purpose devices,
measuring instruments, household devices, and medical devices.

MODELS S P 11  P G 1 0 2

DTerminal Type D : DIP
S : SMD

Rated Pressure

Pressure Type PG : Gauge ()VG : Gauge ()CG : Gauge ()EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT

12SMD Type

3Unit : mm Tolerance ± 0.3mm
123No. Signal Name
Name
Vcc
Power Input
12
NC
Unconnected
Output Voltage
Vout3
UnconnectedNC
4
Unconnected
NC
5
GND
GND
62.2

2.2

Vent Hole

Vent Hole54

3.0

654

5.0

3.0

2.0

8.5

2.0

8.5

5.06

Recommended Circuit(9.5)

7.0 max.
2.5 2.5

2.5 2.5
7.2 max.

7.5

7.2 max.

6-0.90.5

2.5 2.5

Vcc

VoutC1C29.5

7.0 max.
2.5 2.5

0.5

10.0

0.25

0.15

(5.0)

0.25

1.9

DIP Type

1.1
0.5

Recommended hole drillingfor printed circuit boards

GND
C1=0.1μF
C2=1000pF

Recommended padfor printed circuit board

SPECIFICATIONSCG102Model No.
Terminal Type○○D (DIP)
S (SMD)

Rated Pressure
Maximum Overload Pressure
Pressure Type
Pressure Medium
Operating Temperature Range
Compensation Temperature Range
Supply VoltageOutput Voltage *1,2
Total Accuracy *1,2
Maximum Current Consumption *1[ kPa ]

100

VG102

○
○

100

[ C ][ C ][V][V][ %FS ][ mA ]

PG501

○

PG102

○
○

PG202

○
○

No Description
50
100200
Rated Pressure  2Gauge
Non-Corrosive Gases
20 ~ 85 (No Freezing and Condensation)
0 ~ 60 (No Freezing and Condensation)
50.25

PG502

○
○500

PG103

○
○

1000
Rated Pressure  1.5

0.5 ~ 4.5
1.25
10NOTE *1 Shows the value when the supply voltage is 5V. The value fluctuates due to fluctuations in voltage, but the error is not included.
*2 The total accuracy is the factory value. Please design so that it can be corrected when the offset voltage deviates.

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-ssp_series-03

Pressure Sensor Non-influenced by Noise / SPAN Series
FEATURES
Can be used in environments where high-frequency noise occurs.
● Insulation separation structure is adopted in the sensor case.
● High output and high reliability achieved through combination ofmetal diaphragm and thin film gauge.
● High insulation.
● Wide operating temperature range.
● Small size.

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT

Signal Name

OutputPower SupplyRedGNDBlackOutput

WhiteDimensions:AA
58±268±2

M terminal
7114-1260
M housing
Dedicated Housing
F terminal
7116-1261
F housing7123-1835
Waterproof Stopper 7157-3596-30

SPECIFICATIONS

Case

Model No.

SPAN-18.6 0.1
015±1°258 connector V Type Part No.

13 or more

G3/8

HEX24

28

2.5 0.4
0GND

Specifications
Voltage Output
Free Power Supply /Current Output

Wire Color

18 or more

15.3±0.5

A

Power Supply

6 .3 zWire Color

Terminal Arrangement

O RingUnit : mm

Recommended Dimensions
for Sensor Mounting HolesG3/8-Class B

9Structure to Isolate Insulation

Element Part-5  10  20  35  50 MPa

Rated Pressure

150%FS

Overload
Operating Temperature range±1.0%FS or less
40 ~ 120 C (at DC 5V)

Insulation Ring

20 ~ 85 °C

Compensation Temperature range
Power Supply

DC 5V  12V  24V  8 ~ 35V

Output

0.5 ~ 4.5V  1 ~ 5V  4 ~ 20mA

Zero temperature Characteristics

Within ±0.025%FS/°C

Span temperature Characteristics

Within ±0.025%FS/°C

Sensor Output Fluctuation Rate(% FS)

Total Accuracy

2 ms or lessResponse Time

About 60 g

WeightIP67equivalent

IP Protection Rating

Total Accuracy = √ (Linearity) 2 + (Hysteresis) 2 + (Reproducibility) 2

MODEL NUMBER CONFIGURATION

1SPAN
RatedPressure

055 MPa

1010 MPa2020 MPa

3535 MPa

5050 MPa

2Chalk200

Sensor Output Fluctuation Due to Noise

150

Conventional Product
Noise Resistance Product100500500.1k

1k

10k100k*3Output Format

Input Voltage
Voltage Output Specifications
Current Output Specifications

55V

612 V

08 ~ 35 V(Free Power Supply)

724 V

Ex.1) 10 MPa Input Voltage 12 V Voltage Output (0.5 to 4.5 V)
Ex.2) 35 MPa Free Power Supply 24 V Voltage Output (0.5 to 4.5 V)
Ex.3) 50 MPa I nput Voltage 24 V Current Output
Ex.4) 5 MPa Free Power Supply and Voltage Output (1 to 5 V)

h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >H

0.5 ~ 4.5 V

G

1~5VI

4 ~ 20 mA

Voltage Output
Current Output→ SPAN-10-6H→ SPAN-35-0H→ SPAN-50-7I→ SPAN-05-0G

Specifications of products are subject to change without notice.

<1M

Applied Noise Frequency (Hz) 10VP-P Applied

3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-span-p01-03

Pressure Sensors

TAISEI Co.,Ltd.
HEAD OFFICE 6972, Shimo-yoshida, Chichibu-shi, Saitama-ken, 369-1593, Japan.

TEL:81-494-77-1211 FAX:81-494-77-1415( BUSINESS )MKT-TAISEI Co.,Ltd.
HEAD OFFICE 3-10-8, Nohagi 1st Bldg, Minami-ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.

TEL:81-3-3982-0029 FAX:81-3-3986-1537
E-mail : info@mkt-taisei.com
KANSAI BRANCH 3-18-9, Nishi-nakajima, Yodogawa-ku, Osaka-shi, 532-0011, Japan.
TEL:81-6-6390-7264 FAX:81-6-6390-6945

URL : http://www.mkt-taisei.co.jp/2024.09

57