OJ ACNT O
U AB RE RY

2023

OR T HOPEDIC INS T RUMEN T S®

Stoll Bone Plate Clamp

Modified Winged
Fukuda Retractor

Page 27

Page 7

Featuring
many

instruments
insideWiater Shoulder Slide
Page 2Kolbel Self-Retaining
Glenoid Retractor withCenter Blade & Ergonomic
HandlePage 18

Chandran Distal Biceps Tissue Protector
Page 32

Shoulder & Elbow Instruments
1.800.548.2362 • INNOMED.NET

What’s New In This Catalog?
a snapshot of all the

instruments within
Chandran Distal Biceps
Tissue Protector

Bacastow Arthroscopic
Deltoid Lift Retractor

Page 32

Beicker Hammerhead
RongeurPage 33

Page 21

Superior Coracoid
Retractor
Page 8

OrthoLucent™ Modified
Fukuda-type Retractors
Page 7

Rogozinski Glenoid
Retractor

Wiater Shoulder Slide

Page 9Page 2

Wiater Shoulder Slide
Designed by J. Michael Wiater, MD, FAAOS, FAOA

Designed to help avoid damage to the prosthetic
bearing surfaces during dislocation and reductionof a shoulder arthroplasty

Also useful for total hip arthroplasty or hip preservation procedures insmaller patients. Manufactured of delrin to help eliminate damage tothe implant. Can be steam or gas sterilized and is radiolucent.
PRODUCT NO:

6879

Overall Length: 11" (27,9 cm)
Width: 1.375" (3,5 cm)

2

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Shoulder Surgery
Retractor
SystemDeveloped in collaboration with Mayo Clinic.
PRODUCT NO’S:

Modified Thin Glenoid
Retractor–Narrow & Wide

Right Angle
Hohmann Retractor

1251-00 [Complete System]
Included in Set/Available individually:

1252-N [Modified Thin Glenoid Retractor–Narrow]
Two included in set; one with this product number
Overall Length: 11.875" (30,2 cm)
Blade Width: 15 mm

1252-W [Modified Thin Glenoid Retractor–Wide]Two included in set; one with this product number
Overall Length: 11.875" (30,2 cm)
Blade Width: 23 mm

Modified Fukuda
Retractor

1253 [Right Angle Hohmann Retractor]
Overall Length: 8.125" (20,6 cm)
Depth from Bend: 4.25" (10,8 cm)
Blade Width: 16 mm

1254 [Modified Fukuda Retractor]
Overall Length: 8.625" (21,9 cm)
Depth: 2.75" (7 cm)
Blade Width: 39 mm

1255-L [Brown Deltoid/Richardson Retractor–Large]

Brown Deltoid/Richardson
Retractor–Large & Small

Overall Length: 10.5" (26,7 cm)
Depth: 2.5" (6,4 cm)
Blade Width: 60 mm

1255-S [Brown Deltoid/Richardson Retractor–Small]
Overall Length: 10.5" (26,7 cm)
Depth: 2.5" (6,4 cm)
Blade Width: 44 mm

1256 [Modified Darrach Retractor, Straight–Narrow]
Overall Length: 10.25" (26 cm)
Blade Width: 12,7 mm

1257 [Modified Darrach Retractor, Straight–Wide]

Modified Darrach Retractor,
Straight–Narrow & Wide

Overall Length: 10.25" (26 cm)
Blade Width: 19 mm

1258 [Modified Darrach Retractor, Bent–Narrow]
Overall Length: 10.75" (27,3 cm)
Blade Width: 12,7 mm

1259 [Modified Darrach Retractor, Bent–Wide]
Overall Length: 10.75" (27,3 cm)
Blade Width: 19 mm

Modified Darrach Retractor,
Bent–Narrow & Wide

1260 [Soft Tissue Shoulder Retractor]
Overall Length: 10" (25,4 cm)
Depth from Bend: 3" (7,6 cm)
Blade Width: 19 mm

1261 [Glenoid Access Retractor]

Soft Tissue
Shoulder Retractor

Overall Length: 13.5" (34,3 cm)
Access Hole Internal Diameter: 36 mm X 30 mm
Depth of Prongs: 8.5 mm

System includes two of each size of the Modified Thin
Glenoid Retractors, and one of each of the other retractors.

Glenoid Access Retractor

Wiater Shoulder Bone Hook
Designed by J. Michael Wiater, MD, FAAOS, FAOA

Large bone hook designed to retract the proximal
humerus posteriorly to help provide exposure for glenoid
reaming during open shoulder procedures
PRODUCT NO:

Also useful for other large joint procedures.

5079

Curve Diameter: 2.68" (68 mm)
Overall Length: 11.25" (28,6 cm)
Handle Width: 4" (10,2 cm)

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

3

Bell-Hawkins Shoulder
Frame and Blade Set
Designed by Robert H. Bell, MD and Richard Hawkins, MD

Retractor and Frame System for Total
Shoulder Arthroplasty
Deep Adjustable Retractor
Medium Adjustable Retractor

PRODUCT NO’S:

4696-00 [Bell-Hawkins – Complete Set]
Included in Set/Available individually:

Small Adjustable Retractor

4696-01 [Small Adjustable Retractor]
Overall Length: 7.375" (18,7 cm)
Handle Length: 6" (15,2 cm)
Blade Width: 1.25" (32 mm)
Blade Depth: .8" (20 mm)

4696-02 [Medium Adjustable Retractor]
Overall Length: 7.375" (18,7 cm)
Handle Length: 6" (15,2 cm)
Blade Width: 1.7" (43 mm)
Blade Depth: 1.25" (32 mm)

4696-03 [Deep Adjustable Retractor]
Overall Length: 7.375" (18,7 cm)
Handle Length: 6" (15,2 cm)
Blade Width: 1.6" (41 mm)
Blade Depth: 2" (51 mm)

4696-04 [Small Fixed Retractor]
Overall Length: 3" (7,6 cm)
Handle Length: 1.5" (3,8 cm)
Blade Width: 1.25" (32 mm)
Blade Depth: .8" (20 mm)

4696-05 [Medium Fixed Retractor]
Overall Length: 3" (7,6 cm)
Handle Length: 1.5" (3,8 cm)
Blade Width: 1.7" (43 mm)
Blade Depth: 1.25" (32 mm)

4696-06 [Deep Fixed Retractor]
Overall Length: 3" (7,6 cm)
Handle Length: 1.5" (3,8 cm)
Blade Width: 1.25" (32 mm)
Blade Depth: 2.375" (60 mm)

Small
Fixed Retractor

MediumFixed Retractor

Deep
Fixed Retractor

4696-07 [Adjustable Lock Block]

Dimensions: 1.375" x 1" x .85" (35 mm x 25 mm x 20 mm)

4696-Frame [Frame Assembly]

Dimensions: 10" x 9" (25,4 cm x 22,9 cm)

Frame Assembly

Adjustable Lock Block

Bacastow Axillary Nerve
Retractor with Suction
Designed by David Bacastow, MD

Designed with a curved tip to slip all theway under the capsule during shoulder
surgery, helping to protect the axillary
nerve, while also providing suction ofsmoke away from the surgical site

PRODUCT NO:

8739

Made of autoclavable Radel material, the unit is nonconductive of current and resists the high temperaturesassociated with the use of electrocautery.

4

Overall Length: 11" (27,9 cm)
Width: .5" (12,7 mm)
Tongue Length: 1.2" (30 mm)

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Hawkins Shoulder
Instruments
Designed by Richard J. Hawkins, MD

Large Spreader
w/Articular Arms

Designed to enhance
exposure during shoulder
arthroplasty procedures

Small Spreader
w/Articular Arms

PRODUCT NO’S:

5090 [Small Spreader w/Articular Arms]
Overall Length: 6.25" (15,9 cm)
Arm Depth: 2.25" (5,7 cm)
Prong Width: 21 mm
Prong Length: 16 mm

5091 [Large Spreader w/Articular Arms]

Anterior
Capsular
Retractor

Small
Pectoralis
Retractor
Extra Small
Pectoralis Retractor
Cobb Elevator

Overall Length: 10.5" (26,7 cm)
Arm Depth: 2.375" (6 cm)
Prong Width: 23 mm
Prong Length: 23 mm

5092 [Anterior Capsular Retractor]
Overall Length: 11.25" (28,6 cm)
Handle Length: 5.25" (13,3 cm)
Blade Depth: 3.25" (8,3 cm)
Blade Width: 19 mm

5093 [Small Pectoralis Retractor]
Overall Length: 10.25" (26 cm)
Handle Length: 5.25" (13,3 cm)
Blade Depth: 2.5" (6,4 cm)
Blade Width: 25 mm

5094 [Extra Small Pectoralis Retractor]
Overall Length: 11" (27,9 cm)
Handle Length: 5.25" (13,3 cm)
Blade Depth: 1.5" (3,8 cm)
Blade Width: 25 mm

5095 [Cobb Elevator]

Overall Length: 11" (27,9 cm)
Handle Length: 5.5" (14 cm)
Blade Width: 19 mm

5096 [Humeral Head Retractor]
Overall Length: 9" (22,9 cm)
Blade Depth: 2.75" (7 cm)
Blade Width: 37 mm

HumeralHead
Retractor

5097 [Anterior Glenoid Retractor]
Overall Length: 11" (27,9 cm)
Blade Depth: 2.75" (7 cm)
Blade Width @ Fat Pad: 34 mm
Blade Width @ Neck: 18 mm

5098 [Deltoid Retractor]
Overall Length: 9.5" (24,1 cm)
Blade Depth: 3.75" (9,5 cm)
Blade Width @ Fat Pad: 45 mm
Blade Width @ Neck: 32 mm

Anterior Glenoid
Retractor

Deltoid
Retractor

5099 [Modified Darrach Retractor]
Overall Length: 10.75" (27,3 cm)
Blade Width: 19 mm

Modified
Darrach Retractor

Axillary Nerve Protector
Designed by Brett Sanders, MD

Designed for inferior capsular releaseduring shoulder arthroplasty and glenoid exposure
The tapered freer end helps separate the axillary nerve and inferior capsule, even indifficult exposures. Non-conductive material allows the use of a bovie knife directlyin the small channel cutting guide (on both sides). Reversible for right and left use.

1.800.548.2362       OCTOBER 2023   

PRODUCT NO:

8029

Overall Length: 7.125" (18,1 cm)
Width: 12 mm
Thickness: 4 mm   SHOULDER INSTRUMENTS

5

Shoulder Instruments

Designed by Evan Flatow, MD& Louis Bigliani, MD

Modified Darrachtype Elevators

Available in four widths. Used for soft
tissue retraction and exposure. May
also be used to lever the humeral headinferiorly or superiorly and medially to
expose the humeral head from the glenoid
while dislocating the humeral head after
subcapularis removal. May also be used
to retract the humeral shaft posteriorly tohelp expose the glenoid.

Thin Glenoid
Retractors

Available in narrow and wide sizes.
Used for retraction of the anterior
and posterior aspects of the anterior
and posterior glenoid rim.

Posterior Glenoid
Elevators

Available in three widths. Used to help
expose the posterior aspect of the glenoid.
The curved tip allows the elevator to fit on
the posterior rim of the glenoid. The curvein the elevator contours to the humeral
shaft for posterior retraction.

Spiked Darrachtype Elevator

The spiked elevator is used slightly below
the anterior rim of the glenoid to help
retract the labrum and anterior capsule.

Bicep Elevator

Used to help retract the biceps tendon
superiorly. The two extensions allow thelong head of the biceps to fit between
them. The edges fit on the superior
portion of the glenoid rim.

Modified Fukudatype Retractors

Available in small and large sizes.
Used to retract the humeral shaft
posteriorly and helping to expose
the entire glenoid surface.

6

WWW.INNOMED.NET   PRODUCT NO:

1900 [Complete Set]

PRODUCT NO’S:

1950 [3/8" (10 mm)]Blade Width: 10 mm
Overall Length: 10.75" (27,3 cm)

1955 [1/2" (13 mm)]Blade Width: 12 mm
Overall Length: 10.75" (27,3 cm)

1960 [3/4" (19 mm)]Blade Width: 19 mm
Overall Length: 10.75" (27,3 cm)

1965 [1.0" (25 mm)]Blade Width: 25 mm
Overall Length: 10.75" (27,3 cm)

PRODUCT NO’S:

1910 Narrow

Blade Width: 14 mm
Overall Length: 11" (27,9 cm)

1920 Wide

Blade Width: 22 mm
Overall Length: 11" (27,9 cm)

PRODUCT NO’S:

1980 [3/8" (10 mm)]Blade Width: 10 mm
Overall Length: 11" (27,9 cm)

1985 [1/2" (13 mm)]Blade Width: 12 mm
Overall Length: 11" (27,9 cm)

1990 [3/4" (19 mm)]Blade Width: 19 mm
Overall Length: 11" (27,9 cm)

PRODUCT NO:

1970 Narrow

Blade Width: 19 mm
Overall Length: 10.75" (27,3 cm)

PRODUCT NO:

1975

Blade Width: 25 mm
Overall Length: 10.75" (27,3 cm)

PRODUCT NO’S:

1930 Narrow

Blade Width: 32 mm
Opening: 25 x 40 mm
Overall Length: 7.25" (18,4 cm)

1940 Wide

Blade Width: 38 mm
Opening: 32 x 40 mm
Overall Length: 7.25" (18,4 cm)   FREE TRIAL ON MOST INSTRUMENTS

OrthoLucent™ Modified Fukuda-type Retractors

Used to retract the humeral shaft posteriorly, helping to expose theentire glenoid surface, the carbon fiber PEEK composite material
is strong, lightweight, completely radiolucent, helps to prevent frommarring component surfaces, and can be steam sterilized
PRODUCT NO’S:

1930-R [OrthoLucent™ Narrow]
Blade Width: 32 mm
Opening: 25 x 40 mm
Overall Length: 7.5" (19,1 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

1940-R [OrthoLucent™ Wide]
Blade Width: 38 mm
Opening: 32 x 40 mm
Overall Length: 7.5" (19,1 cm)

OrthoLucent™ Wide

Modified Fukuda-type Retractor
with Reamer Slot
Designed by Richard J. Miller, MD
PRODUCT NO’S:

1898 [Narrow]

Overall Length: 7.25" (18,4 cm)
Blade Width: 32 mm
Opening: 25 x 40 mm

1899 [Wide]

Overall Length: 7.25" (18,4 cm)
Blade Width: 38 mm
Opening: 32 x 40 mm

Center cutout slot allows the shaftof a reamer to fit more posteriorly

Evans Modified Fukuda-type Retractors
Designed by Peter J. Evans, MD

Designed to retract the humeral shaft
posteriorly, helping to expose the glenoid surface
Center groove allows a reamer shaft to fit more posteriorly.
PRODUCT NO’S:

5180-N [Narrow]

Overall Length: 8.625" 21,9 cm)
Blade Width: 1" (25.4 mm)
Blade Depth: 3.75" (9,5 cm)

5180-W [Wide]

Overall Length: 8.625" 21,9 cm)
Blade Width: 1.25" (31,7 mm)
Blade Depth: 3.75" (9,5 cm)

Modified Winged Fukuda Retractor
Designed by Scot Rheinecker, PA

Designed with flared edges for lesspressure on soft tissues
PRODUCT NO:

1896

Overall Length: 7.5 (19,1 cm)
Blade Width: 36 mm
Opening: 29 x 40 mm

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

7

Horseshoe Shoulder Frame and Blade Assembly

Designed to enhance exposure during shoulder arthroplasty procedures

Horseshoe Frame

Horseshoe Blade – Wide

Horseshoe Blade – Narrow
PRODUCT NO’S:

2030-00 [Set]

Includes the frame and the five blades

Also available individually:2030-01 [Horseshoe Frame]

Overall Dimensions: 7" x 5" (17,8 cm x 12,7 cm)
Frame Width: .7" (15 mm)

2030-02 [Blade – Wide]

Horseshoe Blade – Narrow, Bent

Blade Width: 22 mm
Overall Length: 11" (27,9 cm)

2030-03 [Blade – Narrow]
Blade Width: 14 mm
Overall Length: 11" (27,9 cm)

Supero LateralApproach

2030-04 [Blade – Narrow, Bent]

Horseshoe Blade – Small Narrow

Blade Width: 14 mm
Overall Length: 10" (25,4 cm)
Handle Length: 4.5" (11,4 cm)

2030-05 [Blade – Small Narrow]
Delto Pectoral
Approach

Blade Width: 16 mm
Blade Depth: 2"
Overall Length: 8.5" (21,6 cm)

Horseshoe Blade – Inferior

2030-06 [Blade – Inferior]

Blade Width: Outside 34 mm, Inside 24 mm
Overall Length: 11.5" (29,2 cm)
Prong Length: 28 mm

Superior Coracoid
Modification

Superior Coracoid Retractor
Can also be used as a modified minihohmann retractor in small bone surgery.
PRODUCT NO:

1666-02

Overall Length: 6.25" (15,9 cm)

Blount Retractor with Small Handle
PRODUCT NO:

Designed by Ronald Romanelli, MD

4852

A blount retractor with a lightweight ergonomic
handle designed for tissue retraction and closure
assistance in knee, shoulder, and hip arthroplasty

Overall Length: 9.375" (25,1 cm)
Handle Length: 4.625" (11,7 cm)
Blade Depth: 1.5" (3,8 cm)
Blade Width at Widest: .325" (8,3 mm)

Evans Reverse Hohmann Retractor

8

Designed by Peter J. Evans, MD

PRODUCT NO:

Smaller size useful for retracting the deltoid superiorly
or laterally, and also protecting the axillary nerve
inferiorly while simultaneously exposing the glenoid

4547

Overall Length: 8.5" (21,6 cm)
Blade Width: Tapers from 30 mm to 18 mm
Blade Depth: 3” (7,6 cm)
Prong Width: 6 mm

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Rogozinski Glenoid Retractor
Designed by Chaim Rogozinski, MD

Designed with a unique ergonomic profile to help
reduce retraction fatigue and place the assistant's handout of surgical view, while the undersurface helps stabilize
the humeral head to allow excellent visualization of the glenoid
PRODUCT NO:

4271

Overall Length: 11.25" (28,6 cm)
Blade Width: .7" (18 mm)
Prongs Width: 3,4 mm
Handle Width: 4.5" (11,4 cm)

Gunther Glenoid Retractor
Designed by Stephen B. Gunther, MD

Ergonomic design helps to retract thehumeral head posteriorly during glenoid
exposure while avoiding reamer contact
during shoulder replacement surgery
PRODUCT NO:

1999

Overall Length: 11" (27,9 cm)
Neck Width: .625" (15,9 mm)
Prong Outside Width: 1" (25,4 mm)
Prong Inside Width: .625" (15,9 mm)

Levy Anterior Glenoid Retractor
Designed by Jonathan Levy, MD

Designed to help alleviate tension on anterior
glenoid structures and the handle is designed
to optionally be clamped to the drape
PRODUCT NO:

4536

Overall Length: 10.5" (26,7 cm)
Depth from Bend: 5.875" (14,9 cm)
Blade Width: .75" (1,9 cm)
Tooth Gap: .325" (8,2 mm)

Modified Darrach-type Bent Elevator
Designed modification by R.L. Stowell, MD of original design by Evan Flatow, MD

Designed for difficult glenoid exposure, theelevator is placed around the posterior glenoid
rim, retracting the cut humeral surface
PRODUCT NO:

1966

Overall Length: 10" (25,4 cm)
Blade Depth: 5" (12,7 cm)
Blade Width: 1" (2,54 cm)

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

9

Capsule Retractors

Retractors for use in Bankart surgery
The single prong retractor is commonly used when retracting
on the inferior rim of the glenoid. The two and three-prong
retractors are designed to be placed medially along thescapular neck to retract the anterior capsule and labrum.
PRODUCT NO’S:

T1008-01 [3 Prongs]

Overall Length: 10" (25,4 cm)
Prong Length: 30 mm

T1008 [2 Prongs]

Overall Length: 10" (25,4 cm)
Prong Length: 30 mm

T1009 [1 Prong]

Overall Length: 10" (25,4 cm)
Prong Length: 30 mm

Deltoid Retractor

Fits easily under the acromion,
deltoid and over the humeral head
Used in most open procedures
PRODUCT NO:

T1001

Width: 30 mm
Overall Length: 8" (20,3 cm)

Humeral Head Retractor

Placed between the glenoid
and the humeral head to obtain
excellent exposure
PRODUCT NO:

T1007

MADE FOR INNOMED IN

GERMANY

Blade Width: 33 mm
Prong Width: 6 mm | 21 mm Gap | 6 mm
Overall Length: 7" (17,8 cm)

Hope Retractor
Designed by Charles A. Hope, MD

Useful in shoulder surgery
PRODUCT NO:

5838

Overall Length: 11" (27,9 cm)
Blade Width: 1" (2,54 cm)

Kirschenbaum
Acromioplasty Retractor
Designed by Ira Kirschenbaum, MD

Helps to protect both the posterior
aspect of the shoulder and the articularsurface of the humeral head during open
acromioplasty and rotator cuff surgery
Designed to fit under the posterior edge of the acromionand lever the humeral head down out of the way.
PRODUCT NO:

5840

Overall Length: 9.25" (23,5 cm)
Blade Width at Tip: 21 mm

10

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Posterior Glenoid Neck
Retractor

Used during osteotomy of the humeral
head and approaches to the glenoid
Designed to allow one finger retraction
Contours to allow teeth to fit behind the glenoid,
retracting tissue for easy access to the glenoid
PRODUCT NO:

T1002

Width: 30 mm
Overall Length: 10" (25,4 cm)

Anterior Glenoid Neck
Retractor

Teeth are specifically designed toretract the subscapularis and capsule
medially during a Bankart procedure
The wide midsection retracts the soft tissue during
anterior glenoid work
The curved handle allows the assistant to use
minimal pressure to achieve exposure
PRODUCT NO:

T1003

Width: 25 mm
Overall Length: 11" (27,9 cm)

Goldstein Glenoid Neck
Retractor

Placed along the glenoid rim
during open Bankart procedure
to allow excellent exposure
The convex teeth sit easily into the glenoid rim while
the strong end of the shaft allows the instrument tostay out of the surgeons view
PRODUCT NO:

T1004

Blade Width at Teeth: 18 mm
Blade Width at Widest: 36 mm
Overall Length: 8.5" (21,6 cm)

Acromioplasty Retractor

Designed to retract and protect thehumeral head during resection of
the inferior acromial surface
The two prongs hook the posterior aspect of theacromion for retraction. The file is used to smoothrough edges of the acromion post-resection.
PRODUCT NO:

S3008

Overall Length: 9" (22,9 cm)
Blade Width: 18 mm

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

11

Angled Glenoid Retractor

Designed by R.L. Stowell, MD

Flared design allows for atraumatic placement
circumferentially about the glenoid — superior,anterior and inferior — during open shoulder
procedures for retraction of the subscapularisand capsule and to facilitate labral work
PRODUCT NO’S:

1901-N [Narrow]

Overall Length: 10" (25,4 cm)
Blade Width at End Tapers from: 1" to .4" (2,5 to 1 cm)

1901-W [Wide]

Overall Length: 10" (25,4 cm)
Blade Width at End Tapers from: 1" to .625" (2,5 to 1,6 cm)

Angled Glenoid Retractor – Forked
Designed by R.L. Stowell, MD

Designed to help with exposure of the difficult
glenoid and facilitation of glenosphere placement
for reverse arthroplasty, with wide and narrow
versions to accommodate most glenoid variations

PRODUCT NO’S:

Specifically designed bend allows the retractor to sit along the posterior
aspect and deliver the glenoid vault into the surgical field while
retracting the humeral head posterior and lateral without excessive
force on the glenoid neck, while the working end of the design allows
reaming and peripheral bone removal without binding on the reamer.

1902-N [Narrow]

Overall Length: 10" (25,4 cm)
Blade Width: 1" (25 mm)
Blade Width at End: .825" (21 mm)

1902-W [Wide]

Overall Length: 10" (25,4 cm)
Blade Width: 1" (25 mm)

Burkhead Glenoid Retractor
Designed by Wayne Burkhead, MD

The retractor bar presses against
the glenoid while the end of theretractor puts pressure on theposterior capsule
PRODUCT NO’S:

5839 [Large]

Overall Length: 9.125" (23,2 cm)
Blade Width at End: 1.5" (3,8 cm)

5839-SM [Small]

Overall Length: 8.75" (22,2 cm)
Blade Width at End: 1" (2,54 cm)

George Semi-Circumferential Glenoid Retractor

Designed by Michael S. George, MD

Designed to depress the humeral headand retract tissue away from the posterior
half of the glenoid, helping to improve
exposure for the preparationand placement of the glenoid
component in total shoulder
arthroplasty
12

WWW.INNOMED.NET   PRODUCT NO:

2435

Overall Length: 8" (20,3 cm)
Blade Width: 2.125" (5,4 cm)   FREE TRIAL ON MOST INSTRUMENTS

Burkhead Reversible TSA/RSA Retractor
Designed by Wayne Burkhead, Jr, MD

Unique shape, angles and double pronged end serves
to push the posterior capsule, and the humerus,
away from the glenoid to allow preparation of
the glenoid and implantation of component(s)without having to remove the retractor
PRODUCT NO:

5839-01

Overall Length: 9.125" (23,2 cm)
Blade Width at End: 1.5" (3,8 cm)

Bacastow Glenoid Retractors
Designed by David Bacastow, MD

Designed for glenoid exposure, particularly
for reverse shoulder replacement applications,where it is important to get inferiorly

PRODUCT NO’S:

1897-L [Left]

Overall Length: 11.75" (29,8 cm)

Allows visualization and direct access to the glenosphere base platethrough a deltopectoral incision with intact pectoralis major insertion.

1897-R [Right]

Overall Length: 11.75" (29,8 cm)

Agrawal Talon Retractor
Designed by Vivek Agrawal, MD

Designed to help facilitate
glenoid exposure in total
shoulder arthroplasty
PRODUCT NO:

4695

Overall Length: 7.875" (20 cm)
Blade Width: 40.7 mm

Glenosphere Component Retractor
Designed by Tim Seachris

Designed for use in total
and reverse shoulder
arthroplasty
Coated version helps
to protect component
surfaces.

PRODUCT NO:

5841 [Coated End]

Overall Length: 10.25" (26,7 cm)
Blade Width: .9375" (2,4 cm)

5841-01 [Uncoated End]

Overall Length: 10.25" (26,7 cm)
Blade Width: .9375" (2,4 cm)

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

13

Weatherly Mini-Deltoid Retractors
Designed by Wallace Weatherly, MD

Designed for the retraction of the deltoid
in a mini-open mid-deltoid splitting
approach to rotator cuff surgery, the offset
handle helps allow clear visualization of
the surgical field, and the ergonomic
non-slip handle surface helps
prevent fatigue in theoperative team

PRODUCT NO'S:

5110-L [Large]

Overall Length: 12.75" (32,4 cm)
Depth from Bend: 4.5" (11,4 cm)
Blade Dimensions: 40 mm x 90 mm

5110-M [Medium]

Overall Length: 11" (27,9 cm)
Depth from Bend: 3" (7,6 cm)
Blade Dimensions: 40 mm x 55 mm

5110-S [Small]

Overall Length: 10.5" (26,7 cm)
Depth from Bend: 2.5" (6,4 cm)
Blade Dimensions: 40 mm x 40 mm

Large
MediumSmall

Rake Retractors
with Ergonomic Handle
Designed for general use
soft tissue retraction

3-Prong

PRODUCT NO’S:

4839 [3-Prong]

Overall Length: 9.5" (24,1 cm)
Rake Width: 13 mm
Rake Depth: 14 mm

4840 [4-Prong]

4-Prong

Overall Length: 9.5" (24,1 cm)
Rake Width: 19 mm
Rake Depth: 14 mm

Meyerding Type Retractors with Ergonomic Handle
75 x 25 mm

Designed for general use soft tissue retraction, the ergonomic
handle allows for a better grip and less fatigue
Non-glare finish featured on the metal retractor parts.

75 x 15 mm

PRODUCT NO’S:

6241 [50 x 16 mm]

Overall Length: 8.875" (22,5 cm)
Blade Width: 16 mm
Blade Depth: 50 mm

6242 [75 x 15 mm]

Overall Length: 9" (22,9 cm)
Blade Width: 15 mm
Blade Depth: 75 mm

6243 [75 x 25 mm]

Overall Length: 9" (22,9 cm)
Blade Width: 25 mm
Blade Depth: 75 mm

50 x 16 mm

Chandler Retractor

PRODUCT NO’S:

3220-01 [5/8"] (15,9 mm)
Overall Length: 9.125" (23,5 cm)
Blade Width: 16 mm

Used for retracting tissue
away from the bone

3220-02 [3/4"] (19 mm)

Overall Length: 9.125" (23,5 cm)
Blade Width: 19 mm

The handle is contoured away fromthe field of view and working area.
Available in three blade sizes: 5/8", 3/4", and 1".
Carbon fiber material is strong, lightweight, completely radiolucent, helps toprevent from scratching component surfaces, and can be steam sterilized.

3220-04 [1"] (25,4 mm)

Overall Length: 9.125" (23,5 cm)
Blade Width: 25.4 mm

3220-02R* [Radiolucent 3/4"] (19 mm)
Overall Length: 9.125" (23,5 cm)
Blade Width: 19 mm* MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

14WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Browne Deltoid
Retractor

Small

Used for the Delto-Pectoral
Approach – the small deep version
is designed for use in large patients

Large
Small Deep

Contours the humeral head with effortless
deltoid retraction allowing extensive exposure.
MADE FOR INNOMED IN

PRODUCT NO’S:

1670-01 [Small]

Blade Width: 45 mm
Overall Length: 11.5" (29,2 cm)

GERMANY

1670-01D [Small Deep]

Blade Width: 45 mm
Overall Length: 12.25" (31,1 cm)

1670-02 [Large]

40 mm deeper thanour standard version

Blade Width: 57 mm
Overall Length: 11.5" (29,2 cm)

Kaminsky OrthoLucent™ Browne-type Deltoid Retractors

Kaminsky
Retractor

Used for the Delto-Pectoral Approach—can remainin place for fracture reduction, plate positioning,and screw/wire/drill location confirmation

Completely radiolucent

Used for acromioplasty, rotator cuff repair, and fracture fixation. Contours thehumeral head with deltoid retraction allowing extensive exposure. Helps to reduce
operative time, assist in fracture reduction, and maintain hardware position without thefrequent need for retractor removal and reintroduction.
Carbon fiber PEI composite material is strong, lightweight, completely radiolucent,can be steam sterilized, and helps to prevent from marring component surfaces.
PRODUCT NO’S:

1670-01R [Small]

Blade Width: 4,5 cm
Overall Length: 10.5" (26,7 cm)

Designed by Sean B. Kaminsky, MD

1670-02R [Large]

Blade Width: 5,4 cm
Overall Length: 10.5" (26,7 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

Levy Wide Deltoid Retractor
Designed by Jonathan Levy, MD

Designed for management of proximal
humerus fractures—facilitates appropriatedeltoid retraction without interference
during active fluoroscopy
Patent Pending
PRODUCT NO:

1672

Overall Length: 11.75" (29,8 cm)
Blade at Widest: 2.5" (6,4 cm)
Blade Depth: 1.375" (3,5 cm)

Contoured to match the curve of the deltoid, the retractor
helps to retract the entire deltoid laterally during thedeltopectoral approach. The width approximates 2/3 the lengthof the deltoid, while the blade is deep enough to help control theentire deltoid without displacement of the tuberosity reduction.
Sized to fit deltoids in small and large patients.

Bolanos Shoulder Retractor
Designed by Alberto Bolanos, MD

PRODUCT NO:

3222

Overall Length: 7.5" (19,1 cm)
Blade Width at Widest: 1" (2,54 cm)

1.800.548.2362       OCTOBER 2023   

Designed for mini-open rotator cuff repairs andshoulder arthroplasty, the contour matches thehumeral head and the rounded edge helps avoid
trauma to surrounding musculature
Depth matches girth of most patients, while the comfortablehandle makes it easier for assistants to hold.

   SHOULDER INSTRUMENTS

15

Mehalik
Posterior Glenoid Retractor
with Long Handle

Designed in collaboration with Mayo Clinic, modified by John Mehalik, MD.

Designed to help expose the posterior
aspect of the glenoid
PRODUCT NO:

1909

Overall Length: 13.2" (33,5 cm)
Access Hole Internal Diameter: 36 X 30 mm

McFarland Shoulder V Retractor
Designed by Edward McFarland, MD

Designed to provide deep access to the
glenoid rim when performing a subscapularis
splitting approach to the shoulder
Fluted to enhace visualization and room when placing suturesin the capsular flaps prior to placing three prong retractors.
PRODUCT NO:

4851

Overall Length: 9" (22,9 cm)
Length To Bend: 8.5" (21,6 cm)
Handle Length: 4" (10,2 cm)
Blade Depth: 2.75" (7 cm)
Blade Width: .625" (1 cm)

McFarland Malleable Shoulder Retractors
Designed by Edward McFarland, MD
PRODUCT NO’S:

4537-00 [Set of Three Sizes]
Also available individually:

4537-01 [Narrow Deep]

Designed to enhance exposurein shoulder procedures

Overall Length: 15.5" (39,4 cm)
Prong Depth: 10 mm

4537-02 [Narrow Shallow]
Overall Length: 15.5" (39,4 cm)
Prong Depth: 6.8 mm

4537-03 [Wide]

Overall Length: 15.5" (39,4 cm)
Prong Depth: 13.5 mm

McFarland
Bent Cobb Elevator
Designed by Edward McFarland, MD

Designed for retraction while
helping to protect the axillary
nerve in shoulder surgery
Ultra hard titanium nitride coating
helps to prolong sharpness.

16

PRODUCT NO:

3431

Overall Length: 9.5" (24,1 cm)
Length from Bend: 3.5" (8,9 cm)
Cobb End Width: .8" (20 mm)
Angle of Bend: 30°

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Bacastow
Shoulder Capsular Retractor
Designed by David Bacastow, MD

Designed to help place tension on theinferior capsule for improved visualizationand dissection when performing anatomic
or reverse shoulder replacement

PRODUCT NO:

5185

Overall Length: 8" (20,3 cm)
Length to Pivot: 5.75" (14,6 cm)
Small Blade Depth: 1" (2,54 cm)
Small Blade Width: 10 mm
Wide Blade Depth: 2" (5,1 cm)
Wide Blade Width: 2.25" (5,7 cm) Maximum
Wide Blade Offset: .75" (1,9 cm)

Rotating arms allow left or right use. Can be usedin conjunction with Axillary Nerve Retractor with
Suction (#8739 – Page 4) for additional safety.

Durham Offset Zelpi Retractor
Designed by Alfred Durham, MD

Staggered depth retractor designed
for exposure during total hip andtotal shoulder surgery
In hip surgery, with the handle towards the surgeon, thelonger leg is on the inside
In shoulder surgery, with the handle downward, the longer leg
is on the outside
The longer leg extends 1.1" (2,8 cm) deeper

PRODUCT NO’S:

1573-L [Left]

Overall Length: 8.5" (21,6 cm)
Leg Depths: 3.1" & 4.2" (7,9 cm & 10,7 cm)

1573-R [Right]

Overall Length: 8.5" (21,6 cm)
Leg Depths: 3.1" & 4.2" (7,9 cm & 10,7 cm)

Small

1.25" Depth

Flat Gelpi Retractors
Medium1.75" Depth

Designed to help retract a broader
area of soft tissue or muscle
PRODUCT NO'S:

4191 [Small]

Overall Length: 6.5" (16,5 cm)
Prong Depth: 1.25" (3,2 cm)

4192 [Medium]

Overall Length: 7.25" (18,4 cm)
Prong Depth: 1.75" (4,4 cm)

4193 [Large]

Overall Length: 9" (22,9 cm)
Prong Depth: 3" (7,6 cm)

The twolargest sizes
feature anergonomic
handle forincreased
comfort andcontrol.

Large

3" DepthDeep5" Depth4194 [Deep]

Overall Length: 10" (24,4 cm)
Prong Depth: 5" (12,7 cm)

Gelpi Retractors
PRODUCT NO’S:

4180 [Standard]Standard

With
Ergonomic
Handle1.800.548.2362       OCTOBER 2023   

Overall Length: 7.75" (19,7 cm)

4181 [With Ergonomic Handle]
Overall Length: 8.75" (22,2 cm)   SHOULDER INSTRUMENTS

17

Kolbel Self-Retaining
Glenoid Retractors

Modified Kolbel Self-Retaining
Glenoid Retractor with Hinge
Two pairs of snap-in, freely pivoting blades included.
PRODUCT NO’S:

T1014-01 [Set – Standard Handle]
T1014-01-2F [Set – Ergonomic Handle]

Standard Handle

Set Includes:

T1015-01 [Retractor – Standard Handle]
Overall Length: 8.25" (21 cm)
Length-to-hinge: 6" (15,2 cm)
Arm Length: 2.25 (5,7 cm)— OR —

T1015-01-2F [Retractor – Ergonomic Handle]
Overall Length: 9.25" (23,5 cm)
Length-to-hinge: 7" (17,8 cm)
Arm Length: 2.25 (5,7 cm)
T1018-P [Blades–Pair] 36 mm X 36 mm
T1019-P [Blades–Pair] 36 mm X 53 mm

Kolbel Self-Retaining
Glenoid Retractor

Ergonomic Handle
For added control and comfort

Standard Handle

Two pairs of snap-in, freely pivoting blades included.
PRODUCT NO’S:

T1014 [Set – Standard Handle]
T1014-2F [Set – Ergonomic Handle]
Set Includes:

Kolbel Self-Retaining Retractor Blades

T1015 [Retractor – Standard Handle]
Overall Length: 8.25" (21 cm)— OR —

T1015-2F [Retractor – Ergonomic Handle]
Overall Length: 9.25" (23,5 cm)

36 x 3636 x 53

36 x 68

36 x 85

T1018-P [Blades–Pair] 36 mm X 36 mm
T1019-P [Blades–Pair] 36 mm X 53 mm

Center blade can be reversed for shallow or deep retraction
Two pairs of snap-in, freely pivoting blades included.
The OrthoLucent™
carbon fiber PEEK blade
is strong, lightweight,
completely radiolucent,helps to prevent frommarring component
surfaces, and can be
steam sterilized.

PRODUCT NO’S:
Wide Blades

T1018 [36 x 36 mm]
T1019 [36 x 53 mm]
T1020 [36 x 68 mm]
T1021 [36 x 85 mm]
Radiolucent Blade

T1019-R* [36 x 53 mm]

Standard Handle

PRODUCT NO’S:

T1050 [Set – Standard Handle]
T1050-2F [Set – Ergonomic Handle]
Set Includes:

T1050-01 [Retractor – Standard Handle]
Overall Length: 8" (20,3 cm)— OR —

T1050-01-2F [Retractor – Ergonomic Handle]
Overall Length: 9" (22,9 cm)

T1050-02 [Center Blade]

Length-to-bend: 6.25" (15,9 cm)
Depth: 2.5" (6,4 cm)
T1018-P [Blades–Pair] 36 mm X 36 mm
T1019-P [Blades–Pair] 36 mm X 53 mm
20 x 53

For added control and comfort

Kolbel Self-Retaining Glenoid
Retractor with Center BladeRadiolucent
Blades20 x 36

Ergonomic Handle

20 x 68

20 x 85

Ergonomic Handle
For added control and comfort

Narrow Blades

T1022 [20 x 36 mm]
T1023 [20 x 53 mm]
T1024 [20 x 68 mm]
T1025 [20 x 85 mm]

Kolbel Self-Retaining Retractor
Two pairs of snap-in, freely pivoting blades included.
PRODUCT NO’S:

T1016 [Set]
Set Includes:

T1017 [Retractor]*

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

18Overall Length: 8.25" (21 cm)
Arm Length: 6.125" (15,6 cm)
Arm Length-to-hinge: 3" (7,6 cm)
T1018-P [Blades–Pair] 36 mm X 36 mm
T1019-P [Blades–Pair] 36 mm X 53 mm

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Two pairs of snap-in, freely pivoting blades included.

T1016-01-2F [Retractor]

Kolbel Self-Retaining
Glenoid Retractor with
Hinge and Double
Ergonomic Handle

Durham Offset Kolbel
Shoulder Retractor Set

PRODUCT NO’S:PRODUCT NO’S:

T1016-01 [Set]
Set Includes:

Overall Length: 9.25" (23,5 cm)
Length-to-hinge: 7" (17,8 cm)
Arm Length: 2.25 (5,7 cm)
T1018-P [Blades–Pair] 36 mm X 36 mm
T1019-P [Blades–Pair] 36 mm X 53 mm

T1030 [Set]

Set includes: (1) T1030-01, (2) T1030-L, (2) T1030-S

Designed for retraction of the deltoid andunder the short head of the biceps muscle
to expose the shoulder, the longer offset
blades are useful in patients with large
muscles, and the shorter offset blades
are useful in smaller elderly patients
Snap-in, freely pivoting smooth curved blades help toconcentrate the forces on the center of the muscle
bellies, allowing the retractor to remain centered andnot get in the way of exposure.

Also available individually:

T1030-01 [Retractor Handle]
Overall Length: 7" (17,8 cm)

T1030-L [Long Offset Blade](2) included in set, (1) only with this product number
Offset Length: 35 mm
Blade Dimensions: 36 x 36 mm

T1030-S [Short Offset Blade](2) included in set, (1) only with this product number
Offset Length: 10 mm
Blade Dimensions: 36 x 36 mm

Designed by Alfred A. Durham, MD

36 x 36 mm
35 mm Offset

36 x 36 mm
10 mm Offset

Kolbel Soft Tissue Retractors
Helps in the early phase to
retract soft tissue comprisingof the gleno-humeral joint
Standard

Use facilitates the introduction of deeper
retractors which are required for sufficient visibilityof the glenoid, acromion and rotator cuff.
PRODUCT NO’S:

T1006 [Standard]

Overall Length: 8" (20,3 cm)

T1006-01 [Deep]

Overall Length: 7.5" (19,1 cm)

Deep

T1006-L [Long]

Overall Length: 8.5" (21,6 cm)

Long

Havens Modified Kolbel
Soft Tissue Retractor
Designed by Philip Havens, MD

Designed for retraction on deltoid split
incisions on mini-open rotator cuff repairs

PRODUCT NO:

T1006-02

Overall Length: 7.5" (19,1 cm)
Opens To: 4.5" (11,4 cm)
Prong Depth: 18 mm

Jaws and arms are parallel with no gap whenclosed to allow easier insertion in tight spaces.

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

19

Deep Meyerding Retractor
with Ergonomic Handle

A self-retaining soft tissue retractor for
use in hip, knee, and shoulder surgery
PRODUCT NO:

MADE EXCLUSIVELYFOR INNOMED IN

6244

GERMANY

Overall Length: 8.5" (21,6 cm)
Blade Depth: 3.5" (8,9 cm)
Blade Width: 1" (2,54 cm)

Subscapularis Spreader

Reaches deep to help split thesubscapularis in a Jobe approach
Also used for retracting a split deltoid in mini
rotator cuff repairs.
PRODUCT NO’S:

T1005 [Standard]

Overall Length: 8.375" (21,3 cm)

T1005-L [Long]

Standard

Overall Length: 9.25" (23,5 cm)

Long

Right Angled Subscapular Spreader – Blunt Tips
Designed by Edward McFarland, MD

Designed to hold the subscapularis muscle open when
performing a subscapularis split approach to the glenoid
PRODUCT NO:

1652

Overall Length: 7.5" (19,1 cm)
Blade Depth: 2" (5,1 cm)

Humeral Protection Plates
Designed by Ronald E. Delanois, MD

50 mm
Spiked Bottom

50 mm
Flat Top

Helps protect the proximal humerus fromfracture after humeral head osteotomy
Plate is placed on the proximal humerus after the
initial osteotomy of the humeral head for total shoulder
replacement. Helps protect the proximal humerus fromfracture as the humerus is retracted to gain visualizationof the glenoid to prepare it for a glenoid implant.

20

PRODUCT NO’S:

5259-01 [46 mm]
5259-02 [50 mm]

46 mm
Spiked Bottom

WWW.INNOMED.NET   

46 mm
Flat Top   FREE TRIAL ON MOST INSTRUMENTS

Gerber Sub-Acromion Spreaders

Designed to gain optimal access to thesubacromion space

Left

Designed to gain optimal access to the subacromion spaceby distracting inferiorly the humeral head from the acromion.
PRODUCT NO’S:
Standard

Modified

1640-01 [Right]

1641-01 [Right]

1640-02 [Left]

1641-02 [Left]

Blade Length: 19 mm
Inside Ring Dia.: 32 mm
Overall Length: 7" (17,8 cm)
Blade Length: 19 mm
Inside Ring Dia.: 32 mm
Overall Length: 7" (17,8 cm)

Blade Length: 34 mm
Inside Ring Dia.: 25 mm
Overall Length: 7" (17,8 cm)

Right

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Blade Length: 34 mm
Inside Ring Dia.: 25 mm
Overall Length: 7" (17,8 cm)

Bacastow Arthroscopic Deltoid Lift Retractor
Designed by David Bacastow, MD

Designed to help improve visualizationof the subacromial space during rotator
cuff repair or balloon spacer placement
PRODUCT NO:

5081

Overall Length: 7.5" (19,1 cm)
Depth from Bend: 2.25" (5,7 cm)
Shaft Diameter: 4,7 mm
Diameter at End: 6 mm

Suprascapular Ligament Cutter
Designed by Michael Craig, OPA-C

Designed to cut the transverse
ligament while helping to protect
the suprascapular nerve
PRODUCT NO:

1794

Overall Length: 11.25" (28,6 cm)

Argintar Biceps Tenodesis Sleeve

PRODUCT NO:

5835

Designed by Evan Argintar, MD

Designed to help facilitate mini-open sub-pectoral biceps tenodesis—
by maintaining the trajectory of the drill with the serrated endof the sleeve, the drilled humeral holes are easily found withstandard percutaneous placement of the bicortical button

Overall Length: 4" (10,2 cm)
External Diameter: 6,35 mm
Internal Diameter: 5 mm
Slot: 2,75 mm

Once flipped, the slotted cut out in the sleeve makes detachment of
the button applicator possible, helping with efficient and reproduciblemini-open biceps tenodesis using button technique.

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

21

Meyer Latarjet Drill Guide & Forceps Assembly
Designed by Professor Dominik Meyer

Aiming device for flush
positioning of a bone blockwith a joint surface
ForcepsDrill Guide

Distance between Drill Holes:
Small = .390" (9,9 mm)
Large = .492" (12,5 mm)

PRODUCT NO’S:
SMALL SETLARGE SET5257-00 [Small Set]

5258-00 [Large Set]

Set Includes:

Set Includes:

5257-01 [Latarjet Forceps, Small]

Overall Length: 5.875" (14,9 cm)
Tongue and Clamp Arm Width: .22" (5.6 mm)

5258-01 [Latarjet Forceps, Large]

Overall Length: 5.875" (14,9 cm)
Tongue and Clamp Arm Width: .32" (8.15 mm)

5257-02 [Latarjet Drill Guide, Small] 5258-02 [Latarjet Drill Guide, Large]
Overall Length: 8.5" (21,6 cm)
Drill Hole Diameter: 3.5 mm
Distance between Drill Holes: .390" (9,9 mm)

1025 [Case]

The osteotomized coracoid
is fixed with the lateral, jointfacing side of the coracoid(where the ligament is) facing
the flange of the drill-guide.

Two 3.5 mm guiding holes
are drilled.

Overall Length: 8.5" (21,6 cm)
Drill Hole Diameter: 3.5 mm
Distance between Drill Holes: .492" (12,5 mm)

Set in Case

1025 [Case]

The drill guide is held against theantero-inferior glenoid, the flange
sitting on the cartilage, and thefirst 2.5 mm thread hole for screw
fixation is drilled.

The second 2.5 mm thread holeis drilled parallel to a 2.5 mm pinthat has been inserted in the firsthole to ensure correct distanceand orientation.

The coracoid is now fixatedusing two 3.5 mm or 4.5 mm
screws flush with the cartilage,
due to the identical distance
between flange and screw holeson coracoid and glenoid.

Coated Inserter for Reverse Shoulder Glenosphere Components
Designed by Michael Radon, Ilya Voloshin, MD, and Nathan Mineo

Designed to aid in the insertion of glenospheres in limited
exposure patients, allowing for insertion from the side, andcoated to help protect marring component surfaces
PRODUCT NO:

5071

Overall Length: 9.5" (24,1 cm)
Inserter Arm Angle: 30°

22

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Burkhead Glenoid Inserter

Designed by Wayne Burkhead, Jr, MD, Michael Radon, and Aaron Merges

Designed to help insert a glenoid component
PRODUCT NO:

4689

Overall Length: 9.875" (25,1 cm)

Glenoid Inserter

Designed by Chase Kuhn & J. Kevin Rudder, MD

Designed for final implantation of the
glenoid prosthesis into the body
Grasping ends are coated to help protectfrom scratching the component surfaces.

PRODUCT NO:

5076

Overall Length: 8.5" (21,6 cm)

Nicholson Headrest
Designed by Gregory Nicholson, MD

Provides excellent exposure to
the shoulder, and patient support

Helps position the patient for
all types of shoulder surgery
in the beachchair position
Designed to provide excellent exposure to theshoulder, the headrest can be used with standard
OR tables (with no modifications to the table).
The headrest provides patient support and helps
position the patient for all types of shoulder
surgery—arthroscopic and open—in the beachchairposition. It can be quickly placed and adjusted.

BEFOREA gel pad forehead strap with velcro is included for optional use.
PRODUCT NO’S:

2450 [Headrest]

Main Plate Dimensions: 6" x 18" (15,2 x 45,7 cm)
Neck Offset Adjustment: 8" (20,3 cm)
Includes/Replacement Parts:
Neck Offset
Adjustment: 8"2450-S [Strap with Gel Pad]
4150-PD2 [Set of 2 Pads]

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

AFTER

23

Resnick Small Bone Tamp with Oblique K-Wire Hole
Designed by Charles Resnick, MD

Design allows for the concurrent reduction of a fracture
and placement of a wire into the fracture site —especially helpful when the surgical exposure issmall and tight, the fracture fragments are
small, and the reduction is demanding

TWO SIZES AVAILABLE:
Wire Hole for K-wires up
to 1.1 mm (.045”) or
1.6 mm (.062”)

BeforeAfter

PRODUCT NO’S:

The serrated distal end minimizes slippage on the cortical surface, does not interferewith the placement of the guidewire and allows for subsequent surgeon-decided,
intraoperative angulation of the wiring once the first cortex is drilled
Especially useful in fractures where there is involvement of an articular surface,for example, mallet fractures of the distal phalanx, articular fractures that involve
ligamentous attachments or tendon attachments of the phalanges, scaphoid polesmall fracture fragments or other small carpal fractures, and radial styloid fractures

5294 [1.1 mm Hole]

Wire Hole for: 1,1 mm (.045") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

5294-01 [1.6 mm Hole]

Wire Hole for: 1,6 mm (.062") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

Stanton Nail/Screw Drill Guide Assembly
FOR DISTAL HUMERAL, FEMORAL, OR TIBIAL SCREWS
Designed by John L. Stanton, MD

Designed to help hold and stabilize a drill guide, allowing thesurgeon to obtain 'perfect circles' and drill distal locking screw
holes without exposure of the hand to the x-ray beam
The drill guide is placed over the side of the bone through an incision. The holder
is attached to the guide and rested against the skin for stability. With the x-ray
on, the guide tube is adjusted by moving the holder until the guide lines up withthe hole in the rod. A drill bit is then inserted into the guide.

PRODUCT NO’S:

Locking Drill Guide Holder

8986-00 [Assembly Set]

Set includes: (1) Holder, (1) Sleeve, and (1) Trocar
Also available individually:

8986-01 [Drill Guide Sleeve]
Overall Length: 3.85" (9,8 cm)
Outside Diameter: 7 mm

Trocar Alignment Tool

8986-02 [Drill Guide Trocar Alignment Tool]

SleeveOverall Length: 4.375" (11,1 cm)
Trocar Diameter: 5 mm

8987 [Locking Drill Guide Holder]
Overall Length: 10.5 (26,7 cm)
Guide Height: 21 mm

24

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Pointed Fracture Reduction Clamps
Designed by Reza Firoozabadi, MD MA

Versatile set of fracture reduction clamps, each witha specific tine design that allows for appropriate
vector placement so that anatomic reduction can beobtained in a number of different types of fractures
1.9 mm tines allow for a snug fit in 2 mm drill holes
Tines angled to prevent clamp “slippage” with compression
Straight tines can be placed deep within bone which allows for far cortex compression.
Clamps incorporate a box joint design that prevents clamp joint loosening and theneed for tightening.
Example applications: any transverse fracture (straight-straight clamp), both bone
forearm fractures, olecranon fractures, medial malleolus fractures, and many more.
Speed Lock Style: Extra-long spin down allows for increased range of clamp use, andopen-topped joint rotates to allow for increased range of opening, and also allows forquick release
PRODUCT NO’S:
SMALL WITH SPEED LOCK MECHANISMMEDIUM WITH SPEED LOCK MECHANISM

3666 [Straight Left & Right]

3666-01 [Straight Left & Right]

3667 [Curved Left & Right]

3667-01 [Curved Left & Right]

Overall Length: 5.5" (14 cm)

Straight Left & Right

Curved Left, Straight Right

Small with
Speed Lock

Small with
RatchetMedium with
Speed Lock

Overall Length: 7" (17,8 cm)

Overall Length: 5.5" (14 cm)

Overall Length: 7" (17,8 cm)

3666-L [Curved Left, Straight Right] 3666-L-01 [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

Overall Length: 7" (17,8 cm)

Overall Length: 5.5" (14 cm)

Overall Length: 7" (17,8 cm)

3666-R [Straight Left, Curved Right] 3666-R-01 [Straight Left, Curved Right]

Two styles — Speed Lock and Ratchet — each available in four tine configurations

SMALL WITH RATCHET MECHANISM

3668 [Straight Left & Right]
Overall Length: 5.5" (14 cm)

3669 [Curved Left & Right]
Overall Length: 5.5" (14 cm)

3668-L [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

3668-R [Straight Left, Curved Right]
Overall Length: 5.5" (14 cm)

Straight
Left & Right

CurvedLeft & Right

Curved Left,
Straight Right

Straight Left,
Curved Right

Durham Bone Reduction Clamps
Designed by Alfred A. Durham, MD

Allows application of a bone plate
without removing the reduction clamp
The large window directly above the jaws provide
space to allow a bone plate to be slid into
position without removing the clamp.

The standard clamp is designed
for medium size bones such as
the fibula, ulna, and radiusStandard

PRODUCT NO’S:

3652 [Standard]

Overall Length: 7.375" (18,7 cm)

3652-01 [Large with Speedlock]
Overall Length: 9.25" (32,5 cm)

Large Window

Provides space to allow a boneplate to be slid into position
without removing the clamp

The large clamp with speedlock
is designed for large bones suchas the femur and tibia

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

Large with
Speedlock25

Calvo Olecranon Reducing Forceps
Designed by Ignacio J. Calvo, MD

Designed to reduce and hold in place transverse
fractures of the olecranon to facilitate theinsertion of k-wires and tension bands
Also very useful in malleolus fractures.
PRODUCT NO’S:

1801-L [Left]
1801-R [Right]

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Durkan Ratchet Bone Clamps
Designed by John Durkan, MD

Design of ratcheting mechanism allows for quicktightening and release around the bone
PRODUCT NO’S:

1867 [Large]

Overall Length: 8.625" (21,9 cm)
Jaw opens to: 3.5" (8,9 cm)

1868 [Small]

Overall Length: 8.5" (21,6 cm)
Jaw opens to: 3.75" (9,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Chen Low Profile Plate/Bone Clamp
Designed by Franklin Chen, MD

Designed for plate to bone clamping in a varietyof lower and upper extremity fractures
Useful for diaphyseal forearm fractures,
humerus fractures, and distal radius fractures.

The rotating nut is used to tighten/
release the center arm once in thedesired position

PRODUCT NO:

1639

Overall Length: 4.5" (11,4 cm)
Prong Depth: 1" (2,5 cm)
End Prong Width: 6 mm|16 mm Gap|6 mm
Clamps from: .375" to 2.4" (1 to 6 cm)

The freely swivelingcenter arm allows foreasy placement, as well
as for quick release, aftergetting the legs in position

Bargo Bone Holding Clamp

Designed to aid in the reduction of various fractures
such as: spiral, transverse, compound, oblique, orbutterfly. The clamp can also be used to secure aplate in place while the screw holes are being drilled
and screws inserted, it has teeth in the jaws for abetter grip, and a ratchet locking handle for use on
various bone diameters. The fracture site can also be
manipulated with the clamp being used as a lever.

Designed by Lonnie Bargo, CST/CFA

Designed to aid in the reductionof various fractures, and canhelp secure a plate in place
during installation

PRODUCT NO:

1895-02

Overall Length: 8.5" (21,6 cm)
Pads: 1.25" x 1" (32 mm x 25 mm)

26

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Fracture Reduction Punch Clamp
Designed by Jong-Keon Oh, MD

Designed for use in select cases when vertical(or sagittal) plane clamping is necessary during
forearm reduction, humeral fracture reduction,or diaphyseal reduction of the tibial shaft
PRODUCT NO:

5072

Overall Length: 10.5" (26,7 cm)
Point to Point Opening:   –Minimum .375" (10 mm)   –Maximum 1.375" (35 mm)
Pin Diameter: .125" (3,2 mm)

Stoll Bone Plate Clamp

Designed by Jordan Stoll, MD

Designed to help hold a bone or bone platein position for reduction and fixation—helpful
with clavicle and fibula fractures

PRODUCT NO:

1774

Overall Length: 10" (25,4 cm)

Browner MIS Bone Clamp
Designed by Bruce D. Browner, MD

Designed to help hold a bone orbone plate for fixation, the clamp isinserted anterior to the bone, rotatedto wrap around the bone, thenscrewed into the desired position
Sized to allow use on a femur, tibia or humerus.
PRODUCT NO:

1379

Overall Length: 9.25” to 11.5”" (23,5 to 29,2 cm)
Maximum Bone Diameter: ~ 35 mm

Pivoting pads accommodate
metaphyseal fractures
The quick release enables adjustment
without losing reduction
Helps provide provisional reduction ofdiaphyseal fractures – humeral shaft
fractures, tibial fractures

Chen Diaphyseal Fracture Reduction Clamp

Designed by Franklin Chen, MD

Designed to facilitate and maintain reductionof the internal fixation of diaphyseal andmeta-diaphyseal fractures of long bones
Works especially well with short oblique bones
while providing room to implement the platewith this bone clamp still in place.

PRODUCT NO:

1808

Overall Length: 9.25" (23,5 cm)
Arm Downward Offset: 15 mm
Pad Dimensions: 1" x .375" (25,4 cm x 1 cm)

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

27

Tissue Graspers
with Shark Teeth

Up Angled Jaw

Designed by Luis Ulloa

Shark teeth help to grasp
on to tissue and bone

Straight Jaw

Shaft allows for use in narrow spaces.
PRODUCT NO’S:

1784-01 [Up Angled Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

1784-02 [Straight Jaw]

Down Angled Jaw

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

1784-03 [Down Angled Jaw]

Shaft Length: 7" (17,8 cm)
Overall Length: 10" (25,4 cm)
Jaw: 9 mm Long x 5 mm High x 1.8 mm Wide

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Cartilage Graspers
Shark Tooth Jaw

Shark Tooth Jaw with 8" (20,3 cm) Shaft

Helps to grasp and holdcartilage, tendons, soft
tissues and loose bodies
Shaft allows for use in narrow spaces.

Saw Tooth Jaw

Shark Tooth Jaw with 5" (12,7 cm) Shaft

Saw Tooth Jaw with 6" (15,2 cm) Shaft

Shark tooth
design modification
by Michael Soudry, MD

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYPRODUCT NO’S:

1777 [5" with Shark Teeth]
Shaft Length: 5" (12,7 cm)
Overall Length: 8.25" (21 cm)
Jaw Bite: 2 mm x 6.5 mm

PRODUCT NO:

1785 [Saw Teeth]

Shaft Length: 6" (15,2 cm)
Overall Length: 9.25" (23,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1779 [8" with Shark Teeth]
Shaft Length: 8" (20,3 cm)
Overall Length: 11.25" (28,6 cm)

Soudry Loose Body Grasper
Designed by Michael Soudry, MD

Designed to help with theremoval of soft tissue loose
bodies in arthroscopy andopen procedures
PRODUCT NO:

1769

Overall Length: 9" (22,9 cm)
Shaft Length: 6" (15,2 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

28

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Shark Tooth Graspers

Sharp teeth help grasp
onto tissue and bone

Designed by Luis Ulloa

Small Grip Handle

Helpful in removing the labrum, and osteophytes around theacetabulum and around the glenoid. Also helps to remove
meniscus, osteophytes and loose bodies. Helps facilitate
working through a small incision without disrupting vision.

9" (22,9 cm) shaft length

PRODUCT NO’S:

1798-SG [7" Small Grip]7" (17,8 cm) shaft length

Jaw Size: 6 mm x 10 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

1799-SG [9" Small Grip]
Jaw Size: 6 mm x 10 mm
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

Small grip version designedwith the grip closer together
for easier gripping and to help
reduce hand fatigue12" (30,5 cm) shaft length

Standard Handle
9" (22,9 cm) shaft length

PRODUCT NO’S:

1797 [5" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 8" (20,3 cm)
Shaft Length: 5" (12,7 cm)

7" (17,8 cm) shaft length

1798* [7" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 10" (25,4 cm)
Shaft Length: 7" (17,8 cm)

5" (12,7 cm) shaft length

1799 [9" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 12" (30,5 cm)
Shaft Length: 9" (22,9 cm)

1796 [12" Standard]

Jaw Size: 6 mm x 10 mm
Overall Length: 15" (38,1 cm)
Shaft Length: 12" (30,5 cm)* MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Powers Modified Kocher Clamps
Designed by Mark Powers, MD

PRODUCT NO’S:

1813 [Tapered Jaw]

Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 5.2 mm x 4.1 mm

Tapered Jaw

1813-01 [Tapered Narrow Jaw]
Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 5.2 mm x 3 mm

1814 [Square Jaw]

Tapered Narrow Jaw

Square Jaw

Overall Length: 8.25" (21 cm)
Jaw Length: 2.5" (6,4 cm)
Jaw at End: 6.5 mm x 5 mm

Heavier design allows for a firmer
grasping of bone and soft tissues

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

29

Hooked Bone Awls

Designed by Reza Firoozabadi, MD“Shoulder hook” awls designed
to help with manipulation ofbone fragments for fixation
PRODUCT NO’S:

5078 [Standard]

Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)

5078-01 [Long]

Standard hooked bone awl being used to gain length
to assist with reduction of a fibula fracture. A 2mm
pilot hole is made to seat the tip of the bone awl.

Overall Length: 13.375" (34 cm)
Handle Length: 6" (15,2 cm)

Lateral Condyle Fracture Set
Designed by Carl R. Weinert, MD

Designed for adult and pediatriclateral condyle fractures

The stops on each clamp end help
prevent excessive penetration of
metaphyseal and soft bone

Elbow Retractor
Designed for use within the elbow joint to
retract the anterior capsule, and provide full
exposure of the anterior articular surface for
reduction and fixation of displaced lateralcondyle fractures. The small blunt tip
hooks over the intact medial condyle.

Symmetric Reduction Clamp
Useful to compress T-condylar fractures, andin many other fracture reduction applications.
The stops on each end help prevent excessive
penetration of metaphyseal and soft bone.

Asymmetric Reduction Clamps – Left and Right
Shaped to secure the lateral condyle fragment. The straight tip isplaced in the coronoid fossa and the curved tip is used to graspand compress the lateral condyle fragment. The stops on each end
help prevent excessive penetration of metaphyseal and soft bone.

PRODUCT NO'S:

4697-00 [Set with Case]
Set Includes / Available Individually:

1755 [Clamp – Symmetric]
Overall Length: 8.5" (21,6 cm)
Jaw opens to: 3" (7,6 cm)

1756-L [Clamp – Asymmetric Left]
Overall Length: 8.75" (22,2 cm)

1756-R [Clamp – Asymmetric Right]
Overall Length: 8.75" (22,2 cm)

4697 [Elbow Retractor]

Overall Length: 6.5" (16,5 cm)
Blade Width: 1" (2,54 cm)

1015 [Sterilization Case]

Dimensions: 11.25" x 7.125" x 3.125" (28,6 cm x 18,1 cm x 7,9 cm)

Beard Distal Bicep Retractor
Designed by David Beard, MD

Designed to help optimize surgical
exposure during anterior single incision
distal biceps tendon reinsertion
The blade design features an anatomically contoured distal end to
hug the radius cortex. The smooth distal end helps to avoid deep
penetration, and the width matches the width of the distal biceps
tendon insertion site. The narrow curved handle design helps to
optimize workspace and visualization.
PRODUCT NO’S:

5834-00 [Set – Retractor & Two Blades]
Available Individually:

5834-01 [Blade] 1 blade with this product number
Overall Length: 6.375" (16,2 cm)
Width: .625" (16 mm)

Sold as a set, or available
individually for replacement5834-02 [Self-retaining Retractor]
Overall Length: 7.5" (19,1 cm)

30

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Auerbach Arm Holder Rake Retractor Set
Designed by David M. Auerbach, MD

Allows intraoperative positioning for proceduresof the posterior arm, elbow, and forearm

Simple design for fast and easy positioning
Connects over the drape in the sterile fieldusing the supplied rail clamp and post
Can be repositioned during surgery
Sterilizable rubber pad protects the arm
Retractors for the skin and soft tissues
connect to the holder
Compact for easy storage

PRODUCT NO’S:

2415-00 [Set]
Individual/Replacement Parts:

2415-01 [Arm Holder Assembly]

Overall Length: 20" (50,4 cm)
Arm Holder Dimensions: 14.5" x 4" (36,9 x 10,2 cm)
Overall Width inclusing Cleats: 7.5" (19,1 cm)

2415-02 [Arm Holder Upright Rod]Overall Length: 19.25" (49,9 cm)

2415-04 [Rake Chain Retractor 4-Prong]
Two included in set, one with this product number
Overall Length including Chain: 10" (25,4 cm)
Retractor Width: .75" (1,9 cm)

2415-06 [Rake Chain Retractor 6-Prong]
Two included in set, one with this product number
Overall Length including Chain: 10" (25,4 cm)
Retractor Width: 1.25" (3,2 cm)

2595 [Table Clamp]
2770-P [Silicone Pad]

Set includes: Arm Holder Upright andAssembly, (2) 4-Prong Chain Retractors,(2) 6-Prong Chain Retractors, (2) Black
Straps, Silicone Pad, and Table Clamp.

Dimensions: 12" x 5.5" (30,5 x 14 cm)

Replacement Parts:

2590-S [Black Straps] Pkg of 10

Distal Humerus Fracture Board
Designed by Burk Young, MD

Designed for the pinning of pediatric supracondylar and adult distal humerus fractures
Allows the surgeon to pin these fractures without having to manually
hold the fracture reduced, allowing the surgeon to focus on accurate pin
placement and reduction. The height of the crossbar is fully adjustable toaccommodate different size patients. Reduction is achieved by an assistantgently applying axial traction through the forearm, with the crossbar
applying the counter traction. Pinning is done with the C-arm in the lateralposition. An optional separate attachment to support the arm for distal
humerus fractures in adults is available. Unit not sterilizable.

Board, crossbar, andoptional adult adapter
are radiolucent.

PRODUCT NO’S:

2445 [Fracture Board – Pediatric]

Main Board Dimensions: 22" x 12" (55,8 cm x 30,5 cm)
Crossbar Height Adjusts From: 4.5" to 7.5" (11,4 cm x 19,1 cm)

2445-01 [Fracture Board – With Adult Adapter]
Optional/Replacement Part:2445-06 [Adult Adapter]

1.800.548.2362       OCTOBER 2023   

Optional

ADULT

Adapter   SHOULDER INSTRUMENTS

31

Chandran Distal Biceps Tissue Protector
Designed by Rama E. Chandran, MD

Designed to help protect tissue and expose theradial tuberosity during distal biceps tendon repair
Using downward pressure, the teeth help to engage bone to keep the protector in place.
Helps to protect tissue and expose the radial
tuberosity during distal biceps tendon repair

Also useful to help expose thehumerous during proximal
subpectoral biceps repair

Also useful to help expose the humerous
during proximal subpectoral biceps repair.
PRODUCT NO:

3224

Overall Length: 8" (20,3 cm)
Handle Length: 5.5" (14 cm)
Blade Depth: 1.7" (4,3 cm)
Blade Width: .65" (1,7 cm)

Vaughan Distal Bicep
Tendon Repair Retractor
Designed by Roderick A. Vaughan, MD

Designed to retract in acontinuous way in three
directions, helping to prevent thesurrounding vital structures fromentering the field while drilling or
performing the repair work
PRODUCT NO:

Profile View

3223

Overall Length: 8.375" (21,3 cm)
Handle Length: 5.25" (13,3 cm)
Depth: 2" (5,1 cm)

Gap Clamp for Cortical Button
Distal Bicep Repair
Designed by Corey Trease, MD

Designed to be used to help
consistently set the gap forthe radius cortex between
the distal biceps stump andthe cortical button

32

WWW.INNOMED.NET   PRODUCT NO:

5262

Overall Length: 8" (20,4 cm)
Jaw Width Tapers from: 6 to 2.5 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   FREE TRIAL ON MOST INSTRUMENTS

Beicker Hammerhead Rongeur
Designed by Clint Beicker, MD

Designed to help remove
osteophytes from around theacetabulum and glenoid
PRODUCT NO:

1775-05

Overall Length: 8" (20,3 cm)
Jaw Bite: 15 mm x 7 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Mazzara Rongeur for Small Bones
Designed by James T. Mazzara, MD

Designed for bone and soft tissue removal
in small joint surgery, the pistol grip handle
lessens hand fatigue and slippage, andallows for better visualization

2 x 10 mm
Jaw Bite4 x 10 mm
Jaw Bite

PRODUCT NO’S:

1765-04

Jaw Bite: 2 x 10 mm
Overall Length: 9" (22,9 cm)

1765-05

Jaw Bite: 4 x 10 mm
Overall Length: 9" (22,9 cm)

Hannum Tissue Grasper2 x 10 mm

4 x 10 mm

Two Jaw Sizes Available

PRODUCT NO’S:

Designed by Scott Hannum, MD

1775-01 [Short Jaw]

Jaw Width: 8 mm
Overall Length: 9.25" (23,5 cm)

Teeth in jaw firmly holds bone and tissue

1775-02 [Medium Jaw]

Non-locking design can be easily gripped while allowing greater pressure
to be applied. Available in three jaw sizes: short jaw for holding bone,
medium jaw for smaller bones, and long jaw for tissue.

Jaw Width: 5 mm
Overall Length: 9.25" (23,5 cm)

1775-03 [Long Jaw]

Jaw Width: 3 mm
Overall Length: 9.25" (23,5 cm)

Jaw widths at actual size

MADE EXCLUSIVELYFOR INNOMED IN

3 mm 5 mm

GERMANY

8 mm

Long Jaw Needle Nose Pliers
PRODUCT NO:

1833

MADE EXCLUSIVELYFOR INNOMED IN

Overall Length: 7" (17,8 cm)
Jaw Length: 2.25" (5,7 cm)
Jaw Width Tapered from: 8 mm to 1.5 mm
Jaw Height Tapered from: 12 mm to 2.5 mm

1.800.548.2362       OCTOBER 2023   GERMANY   SHOULDER INSTRUMENTS

33

STANDARD LARGEPRODUCT NO’S:

OrthoVise™ Length: 10" (25,4 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3980-01 with Attachment Bolts (two sides & end)
without Slap Hammer
3981
without Attachment Bolts
without Slap Hammer
with End Attachment Nut that accepts aStandard Slap Hammer (#3925 or 3926)

3980

#3980

Side Attachment Bolts

Allows a Large OrthoVise™
Slap Hammer (#3950) to be
attached to the side of the device.

#3980-01

Available on Standard Large and Long Nose
Large OrthoVise™ with Attachment Bolts only.
(Cannot be added on later.)#3981

LONG NOSE LARGEPRODUCT NO’S:

OrthoVise™ Length: 12" (30,5 cm)
with Attachment Bolts (two sides & end)
with Large OrthoVise™ Slap Hammer (#3950)
3965-01 with Attachment Bolts (two sides & end)
without Slap Hammer

3965

#3965

OrthoVise™

#3965-01

U.S. Patent #D398,208

LONG NOSE LARGE BENT JAW
PRODUCT NO’S:

OrthoVise™ Length: 11.5" (29,2 cm)
with Attachment Nut (end)with Standard Slap Hammer (#3925)
3966-01 without Slap Hammer
with Attachment Nut (end) that accepts aStandard Slap Hammer (#3925 or 3926)

3966

#3966
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY#3966-01

Made of stainless steel
Models equipped with attachment bolts allow aslap hammer to be attached to the end, as wellas to either side of the large OrthoVise™ (except
Bent Jaw models), for greater adaptability
Bent Jaw models are not available with side
attachment bolts, but have an end attachment
nut to accept a Standard
Slap Hammer (#3925 or #3926)
A different size slap hammer is used for thelarge and small sizes of OrthoVise™
Slap Hammers are designed with a hammer
plate for the additionaluse of a mallet if desired

STANDARD SMALLPRODUCT NO’S:

OrthoVise™ Length: 8" (20,3 cm)
without Attachment Bolt
without Slap Hammer
3985-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3985-T with Attachment Bolt (end)without Slap Hammer

#3985

3985

#3985-01#3985-T

LONG NOSE SMALLPRODUCT NO’S:

OrthoVise™ Length: 9.5" (24,1 cm)
without Attachment Bolt
without Slap Hammer
3975-01 with Attachment Bolt (end)with Small OrthoVise™ Slap Hammer (#3955)
3975-T with Attachment Bolt (end)without Slap Hammer

3975

SLAP HAMMERS#3975#3975-01#3975-T

PRODUCT NO’S:

3950 [Slap Hammer for Large OrthoVise]For use with 3965's, 3980's, 3981
Overall Length: 16.5" (41,9 cm)

For Large
OrthoViseFor use with: 3975's, 3985's
Overall Length: 8.75" (22,2 cm)

For Small
OrthoViseFor use with: 3966's
Overall Length: 16" (40,7 cm)

Standard
with 16" RodFor use with: 3966's
Overall Length: 16" (40,7 cm)

Easy Grip Standard
with 16" Rod3955 [Slap Hammer for Small OrthoVise]3925 [Standard Slap Hammer w/16" Rod]3926 [Easy Grip Slap Hammer w/16" Rod]

#3950

#3955

#3925

#3926

THREADED ADAPTERS
PRODUCT NO’S:

3980-02 [Small Adapter]

Changes Male End of a Slap Hammer to Female

Small Adapter

For use with 3965's, 3966's, 3980's, 3981

Threaded Large

For use with: 3975's, 3985's

Threaded Small

3980-03 [Threaded Adapting Screw – Large]

3985-03 [Threaded Adapting Screw – Small]

34

Female/Female
Adapter Converts
from Male/Male

Small Adapter allows a Standard Slap Hammer (#3925 or #3926)
to be used with any Large OrthoVise™ with Attachment Bolts

Threaded Adapting Screws can be used to append the corresponding size
OrthoVise™ with an Attachment Bolt for use with a Slap Hammer

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Durham Curved Osteotome
Designed by Alfred A. Durham, MD

Increased angle useful for posterior
osteophytes of the humeral head
and the femoral condyle, as well asanterior acetabular osteophytes
PRODUCT NO:

4950

Overall Length: 9" (22,9 cm)
Handle Length: 5" (12,7 cm)
Osteotome Width: .625" (1,6 cm)

Modified Lambotte OsteotomesSix (6) sizes available, from 1/4" to 1-1/2"
in 1/4" increments. Cross-bar and caseincluded in complete set. Two smallest sizes
have an 1/8" hole in which an 1/8" pin canbe used as a cross bar (not included).

Designed with a striking platform, plus across-bar hole to help control rotational
stability and assist with removal
PRODUCT NO’S:

5350-00 [Set w/Case]* MADE EXCLUSIVELY

Also Available Individually:

GERMANY

5350-25* [1/4"]

FOR INNOMED IN

Overall Length: 9" (22,9 cm)
5350-125 [1-1/4"]
Osteotome Width: .25" (6,4 mm)
Overall Length: 9" (22,9 cm)
Osteotome Width: 1.25" (31,8 mm)
5350-50* [1/2"]
Overall Length: 9" (22,9 cm)
5350-150 [1-1/2"]
Osteotome Width: .5" (12,7 mm)
Overall Length: 9" (22,9 cm)
Osteotome Width: 1.5" (38,1 mm)
5350-75 [3/4"]
Overall Length: 9" (22,9 cm)
5350-CASE [Case]
Osteotome Width: .75" (19 mm)
Dimensions: 12.25" x 11.25" x 1"
(31,1 x 28,6 x 2,5 cm)
5350-100 [1"]Overall Length: 9" (22,9 cm)
5350-CB [Cross Bar]
Osteotome Width: 1" (25,4 mm)
Overall Length: 4.375" (11,1 cm)

Mini-lexer Osteotomes

Small, thin osteotomes helpful in osteophyte
and cement removal in total joint surgery.
Larger handle helps with better control.

Helpful in osteophyte and cement removal
PRODUCT NO’S:

5270-01

5270-035270-025270-04

Blade Width: 4 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

MADE FOR INNOMED IN

GERMANY

Blade Width: 6 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

Blade Width: 10 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)
Blade Width: 12 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

PRODUCT NO'S:

5301-00 [Complete Set]
Included In Set / Replacement Parts:

5301-01 [Guide Only]

Overall Length: 5.5" to 8.5" (14 cm to 21,6 cm) w/o blade

5301-02 [10 mm Chisel Blade Only]

WhelanFlexible Chisel Guide

Overall Length: 4.625" (11,7 cm)
Blade Thickness: .020" (0,51 mm)

3040 [Slap Hammer]
1015 [Sterilization Case]

Designed by E. J. Whelan, III, MD

Chisel blade features an ultra hard titanium nitride coating
to help extend life by increasing surface hardness, prolonging
sharpness, and resisting chemicals and corrosion.

Designed to help stabilize a thin chisel blade
until it's within the bone prosthesis interface
Guide with sliding handle helps to stabilize a thin
flexible chisel blade until it's within the bone prosthesis
interface. Chisel tip lets it hug the prosthesis to help
prevent perforation. Slap hammer threads into thehandle and is designed to facilitate blade removal. Easilychangeable disposable blades help assure sharpness.
1.800.548.2362       OCTOBER 2023   

Complete Set
Two blades included in Set   SHOULDER INSTRUMENTS

35

Flexible Osteotome System

Provides an assortment of osteotome blades
for various orthopedic surgery procedures

System Includes
Choice of Handle Style
Handle with Quick-Coupling End

Handle with Locking Nut

PRODUCT NO’S:

S0011-00 [Set with Quick-Coupling Handle and Case]
S0012-00 [Set with Locking Nut Handle and Case]

2.5" Thin Blades

Individual Instruments Included in Sets:

S1002 [Thin Osteotome Blade] 2.5" (6,3 cm) x 8 mm
S1003 [Thin Osteotome Blade] 2.5" (6,3 cm) x 10 mm
S1004 [Thin Osteotome Blade] 2.5" (6,3 cm) x 12 mm
S1005 [Thin Osteotome Blade] 2.5" (6,3 cm) x 20 mm
S1006 [Curved Thin Osteotome Blade] 2.5" (6,3 cm) x 12 mm
S1007 [Curved Thin Osteotome Blade] 5" (12,7 cm) x 20 mm
S1008 [Thin Osteotome Blade] 5" (12,7 cm) x 10 mm
S1009 [Thin Osteotome Blade] 5" (12,7 cm) x 8 mm
S1020 [Handle with Quick-Coupling End] 5" (12,7 cm)

Curved Thin Bladesor5" Thin Blades

S1021 [Handle with Locking Nut] 5" (12,7 cm)
S1133 [Radial Osteotome] 5" (12,7 cm) x 10 mm
S1120 [Radial Osteotome] 5" (12,7 cm) x 12 mm
S1134 [Radial Osteotome] 5" (12,7 cm) x 14 mm
S1121 [Radial Osteotome] 5" (12,7 cm) x 16 mm
S1122 [Radial Osteotome] 5" (12,7 cm) x 20 mm
S2007 [Slap Hammer] 12" (30,5 cm)
9018 [Case]
Sharp, flexible blades are well suited for loosening implants from cement or bony
ingrowth fixation
Various blade widths and profiles allow great flexibility to follow the implant contours
Modular handle is made of high impact surgical stainless steel and has a quickcoupling positive locking mechanism for ease of use and quick blade changes
Slap hammer threads into the handle and is designed to facilitate blade removal

Optional Parts and Blades

Lateral Curve
Radial Blade

Strike Plate for Handle

Optional Parts and Blades
Optional Strike Plate can be attached to the Handle for direct striking with a mallet
Optional Curved Chisel Blades are designed to help loosen the cement/prosthesis
interval in TKA tibial tray and femoral component revisions. The curved design isuseful in working around pegs & fins to get posterior cement access. Also helpful withremoval of other implants, i.e. shoulder, ankle, etc.
PRODUCT NO’S:

S1020-SP [Strike Plate for Handle] Diameter 1.625" (4,1 cm)
Optional Osteotome Blades (Not Included In Complete Set):

S1123 [7.5" XL Osteotome Blade] 7.5" (19,1 cm) x 8 mm
S1135 [Radial Osteo. Medial Curve] 6.75" (17,1 cm) x 11 mm
S1136 [Radial Osteo. Lateral Curve] 6.75" (17,1 cm) x 11 mm
S1137 [Radial Osteo. Medial Curve] 5" (12,7 cm) x 11 mm
S1138 [Radial Osteo. Lateral Curve] 5" (12,7 cm) x 11 mm

Extra Long 7.5" Osteotome BladeMedial Curve
Radial BladeMedial Curve Radial Blades
Curved Radial
Blades are helpful
in the removal of
total hip stems

Lateral Curve Radial Blades

Curved chisel design
allows working aroundcomponent pegs, fins, etc.
2" Curved Chisel Blades

Optional Chisel Blades (Not Included In Complete Set):

S1233-L [2" Left Curved Chisel Bade] 2" (5,1 cm) x 8 mm
S1233-R [2" Right Curved Chisel Blade] 2" (5,1 cm) x 8 mm
S1222 [2.5" Chisel Blade – 8 mm] 2.5" (6,4 cm) x 8 mm
S1223 [2.5" Chisel Blade – 10 mm] 2.5" (6,4 cm) x 10 mm
S1224 [2.5" Chisel Blade – 12 mm] 2.5" (6,4 cm) x 12 mm
S1225 [2.5" Chisel Blade – 20 mm] 2.5" (6,4 cm) x 20 mm
S1229 [5" Chisel Blade – 8 mm] 5" (12,7 cm) x 8 mm
S1228 [5" Chisel Blade – 10 mm] 5" (12,7 cm) x 10 mm
S1231 [5" Chisel Blade – 12 mm] 5" (12,7 cm) x 12 mm
S1230 [5" Chisel Blade – 20 mm] 5" (12,7 cm) x 20 mm
S1227 [5.5" Long Chisel Blade] 5.5" (14 cm) x 8 mm
S1232 [7.5" XL Chisel Blade] 7.5" (19,1 cm) x 8 mm
S1234 [8.5" XL Chisel Blade] 8.5" (21,6 cm) x 8 mm
S1235 [9.5" XL Chisel Blade] 9.5" (23,1 cm) x 8 mm
S1236 [10.5" XL Chisel Blade] 10.5" (26,7 cm) x 8 mm
S1237 [11.5" XL Chisel Blade] 11.5" (29,2 cm) x 8 mm
S1238 [12.5" XL Chisel Blade] 12.5" (31,8 cm) x 8 mm

Radial Blades

Slap Hammer

Bevel side awayfrom component

2.5" Chisel Blades

5" Chisel Blades

5.5" Long Chisel Blade

Extra long chisel blades designed
for removal of well-fixed longbone intramedullary hardware

Blade lengths reflect the actual working portion of the blade only.
For overall length, add 1.5" (3,8 cm) to blade length listed above.
Medial and Lateral Curve Radial Blades designed by Henry Boucher, MD
Curved Chisel Blades designed by William McMaster, MD

7.5", 8.5", 9.5", 10.5", 11.5", & 12.5" Extra Long Chisel Blades

36

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

1
Narrow Cement Removal Gouge, Short
2
Narrow Cement Removal Gouge, Long

Mueller-Type Cement
Removal Instruments

Used for cement removal in
the hip, knee, and shoulder

3
Narrow Offset Cement Removal Gouge

PRODUCT NO’S:

S7500-00 [Complete Set with Case]
4
Acetabular Chisel
5
Offset Chisel
6
Flared Angle Chisel
7
Wide Gouge

Individual Instruments:

S7505 [Narrow Cement Removal
Gouge, Short]
Shaft Length: 15 cm
Gouge: 9 mm, negative

1

Shaft Length: 24 cm
Gouge: 9 mm, negative

2

Shaft Length: 24 cm
Gouge: 9 mm, negative

3

Shaft Length: 24 cm
Chisel: 7.5 mm

4Shaft Length: 15 cm
Chisel: 9 mm

5Shaft Length: 24 cm
Gouge: 9 mm, positive,
angle 15° down

6

Shaft Length: 24 cm
Gouge: 11.5 mm, negative

7

S7507 [Narrow Cement Removal
Gouge, Long]
S7510 [Narrow Offset Cement
Removal Gouge]
S7515 [Acetabular Chisel]
S7520 [Offset Chisel]

8

S7525 [Flared Angle Gouge]"V" Splitter
9

S7530 [Wide Gouge]

Saddle Punch
10
Cement Splitting Osteotome

Shaft Length: 24ccm
Punch: 16.5 mm x 6.5 mm

9
10

12

S7595 [Cement Removal
Osteotome, Short]

11

13

S7597 [Cement Removal
Osteotome, Long]

12

Shaft Length: 24 cm

Cement Removal Osteotome, Long

14
4.4 mm Drill & Drill Guide

Drill

15
16
6.4 mm Drill & Drill Guide

Shaft Length: 15 cm
Osteotome: 8 mm
Shaft Length: 24 cm
Osteotome: 8 mm

S7540 [4.4 mm Drill]
S7545 [4.4 mm Drill Guide]
S7550 [6.4 mm Drill]
S7555 [6.4 mm Drill Guide]
S7560 [Straight Cement
Removal Hook]
Hook Curette: 10 mm

S7565 [Curved Cement
Removal Hook]
Hook Curette: 10 mm

17Straight Cement Removal Hook

Conical Tap & Mallet

8

S7590 [Cement Splitting Osteotome]

Cement Removal Osteotome, Short

Drill Guide

V-Shaped Chisel: 7 mm

S7587 [Saddle Punch]
11

Drill & Drill Guide

S7535 [“V” Splitter]

18

S7570 [Cross Bar]
S7575 [7 mm T-Handle Conical Tap]
S7580 [9 mm T-Handle Conical Tap]
S7585 [Slotted Mallet]
9075 [Case Only]

13
14
15
16
17
18
19
20
21
22

Curved Cement Removal Hook

Conical Tap

19
Slotted Mallet

Optional Instruments:

Cross Bar20
7 mm T-Handle Conical Tap

T-Handle Chuckfor use with Drills

21
9 mm T-Handle Conical Tap

PRODUCT NO’S:

22
Slotted Mallet

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

8247-00 [T-Handle Chuck & Key]8247-01 [T-Handle Chuck Only]
8247-02 [Chuck Key Only]

37

Nicholson Universal
Humeral Prosthesis Extractor

Designed by Gregory Nicholson, MD

Designed to fit most humeral prostheses

Prototype
Shown

PrototypeShown

PRODUCT NO’S:

3670 [Extractor Set with Case]
Individual/Replacement Parts:

3670-01 [Extractor Only]
3670-10 [Foot Adapter]
3670-CABLE [2.5 mm Cable] Package of 2
9007 [Case Only]
3925-A12 [12" (30,5 cm) Slaphammer Rod Only]
3925-H [Slaphammer Only (No Rod)]Set includes a slaphammer,
two non-sterile 2.5 mm cables, and a sterilization case.

Nicholson Shoulder
and Small Bone Cement
Removal Instruments

PRODUCT NO’S:

Gouges Overall Length: 9" (22,9 cm)
Gouges Handle Length: 4" (10,2 cm)

5251-00 [Complete Set w/Case
5251-05 [Extra Small]
Gouge Width: 5 mm

5251-07 [Small]

Designed by Gregory Nicholson, MD

Designed to facilitate cement removal in smaller diameterbone of the humerus, ulna, and smaller implant geometries
Extra Small Gouge
Small Gouge
Medium Gouge
Large Gouge
Small Gouge with Splitter
Medium Gouge with Splitter

Reverse bevel tip helps the gouge
to slide between the bone andcementT-shaped Gouge-Splitter allows the
gouge to slide between the cement
and bone and vertically split
the cement mantle to facilitate
removalSmall diameter widths andcurvatures more closely matchshoulder and elbow implants andsmaller bone diameters
Shorter length allows for better
control and access
The footed impactor is used tohelp remove a humeral implant byimpacting the medial collar of theprosthesis — helps provide a very
direct parallel force to the implant
for removal

Gouge Width: 7 mm

5251-09 [Medium]
Gouge Width: 9 mm

5251-11 [Large]

Gouge Width: 11 mm

5252-07 [Small w/Splitter]
Gouge Width: 7 mm
Splitter Height: 4 mm

5252-09 [Medium w/Splitter]
Gouge Width: 9 mm
Splitter Height: 5 mm

5252-11 [Large w/Splitter]
Gouge Width: 11 mm
Splitter Height: 6 mm

5254 [Backhook]

Overall Length: 12.5" (31,8 cm)
Handle Length: 4.5" (11,4 cm)
Shaft Diameter: 4 mm

5255 [Footed Impactor]

Foot Pad Size: 8.5 mm x 11.5 mm
Shaft Diameter: 8.5 mm (21,6 cm)
Overall Length: 12.75" (32,4 cm)
Handle Length: 4.5" (11,4 cm)

5253 [Case for Set]

Large Gouge with Splitter

Backhook

Prototype
Shown

Footed Impactor

38

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Levy Humeral Stem Extraction Punch

Designed by Jonathan Levy, MD

Ultra hard cobalt chrome shaft and impactor tip designed to help
remove a humeral stem during revision total shoulder arthroplasty
Can be used to open up distal cement mantle or pedestal during revisions.
PRODUCT NO:

8627

Overall Length: 12" (30,5 cm)
Handle/Platform Length: 4.75" (12,1 cm)
Punch Rod Length: 7.25" (18,4 cm)
Platform: 3" x .75" (7,6 cm x 1,9 cm)
Shaft Diameter: 8 mm, tapers to 4 mm at tip

Nicholson Footed Impactor
Designed by Gregory Nicholson, MD

Designed to help remove a humeral
prosthesis by impacting the medialcollar from underneath, after a gap
has been exposed between the rim/bone interface
PRODUCT NO:

5255

The distal, footed end of the impactor is positioned under the neck rim of
the prosthesis, and a mallet is used to strike the large proximal platform of
the impactor to help loosen and remove the prosthesis in line with the stem.

Foot Pad Size: 8.5 mm x 11.5 mm
Shaft Diameter: 8.5 mm (21,6 cm)
Overall Length: 12.75" (32,4 cm)
Handle Length: 4.5" (11,4 cm)

Wagner Osteotome Handle
Designed by Russell Wagner, MD

Handle is designed for easier gripping, rotational
control, and use with a mallet with a standard
1/4” Lambotte osteotome

PRODUCT NO'S:

5348 [Handle Only]

Overall Length: 5.5" (14 cm)

5348-01 [1/4" Osteotome Only]
Overall Length: 8.875" (22,5 cm)

Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

1.800.548.2362       OCTOBER 2023      SHOULDER INSTRUMENTS

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:

39

PRSRT. STD.
U.S. POSTAGE
PAID
GRAND RAPIDS, MI

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 748

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

Dodson Modular Retractor
Designed by Mark A. Dodson, MD

Allows the limb to be rotated (pronated or supinated) without lossof exposure. The hohmann retractors have three hole sizes whichallow for a variety of positioning angle options using the teeth of theself-retaining retractor, or can also be positioned in-between theteeth. The hohmann is placed around the bone, and thus reduces
the force on the soft tissues while increasing exposure. Can beused in the forearm to treat radius and ulna shaft fractures,
humerus fractures, as well as in the leg for fibula fractures.

Designed to help expose a small to medium sizebone for internal fixation—can be used for humerus,
distal radius, ulna, and fibula fractures

PRODUCT NO’S:

Set consists of
one ratcheting selfReplacement Parts:
retaining retractor,
two stainless steel
1838-01 [Retractor Only]
Overall Length: 5.5" (14 cm)
mini-hohmann
1838-02 [Blade Only – One] retractor blades, anda sterilization case.
Overall Length: 5.25" (13,3 cm)
1838-00 [Set]

Blade Width: 3/8" (9 mm)

1025 [Sterilization Case Only]
Optional Parts — Not Included In Set:

1838-02R* [Radiolucent Blade Only – One]
Overall Length: 5.25" (13,3 cm)
Blade Width: 3/8" (9 mm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY* MADE EXCLUSIVELYFOR INNOMED IN

US Patent No. 9,161,745 B2

SWITZERLAND

Optional radiolucent carbon fiber PEEK composite blade
The optional radiolucent blade is made of a strong, lightweight carbon fiber PEEK
composite material, which is completely radiolucent, helps to preventfrom marring component surfaces, and can be steam sterilized.

INNOMED, INC

Innomed-Europe LLC

TOLL FREE 1.800.548.2362

Tel 912.236.0000
Fax 912.236.7766

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74

Innomed-Europe GmbHISO 13485:2016

103 Estus Drive
Savannah, GA 31404

Villingen-Schwenningen,Deutschland
Tel 0049 (0) 7720 46110 60

innomed.net
info@innomed.net© 2023 Innomed, Inc.
All Rights Reserved

10
23

www.innomed-europe.com
orders@innomed-europe.com

INNOMED.NET

info@innomed.net