NAKAYAMACo.,Ltd.
For On-Floor Melt Control

QCR-WinQCR Series Windows Edition

Data access through USB
Data Storage and Review
Touch Panel

CE modeSG modeThermal Analysis for CE, C%, and Si%
Charge Calculation Initial and addition

Nodularity (SG%) ofDuctile and CV cast ironFeatures of QCR-Win
Simplified
calibration curve
preparation

Improved precisionof temperature
measurement

Image preservation
function of themeasurement result

Feature

Can be recorded and management of temperature data.
The measurement data can be extracted with a simple operation (USB memory).
The measurement data can be managed at the office PC.

optionPrint function to the printer.
Output function to the display device.
Data transfer function of a LAN.

Recording and management of temperature data
Communication method:

thermometer ; wireless Receiver; RS232

Use models (selling separately):
TMW-2000 / TMW-3000 / TMW-ZERO / TM-RW

Immersion thermometer is possible one.
(Up to three)

Thermometer system examples, please refer tothe catalog of wireless thermometer

System example

A

Displayed on a large display unit
Communication method: Wire communication
Use models: 3-stage indicator
3-stage indicator

BPrinter output
Communication method: USB、RS232

Print contentdate

TM-PR

Data copied to a USB memory

CUse models: TM-PR

2016 / 2 / 10timeCHCE

S%

Si% …

00:00 ch1 4.00 3.40 1.80

Storage to the internal server
Use models: Cable LANNAS

DAchievement control
Communication method: Manually copy(USB) or NAS

Wired and wireless communication

Office PC*Please contact us for wireless of wired part.

Network system※Need a LAN port additional to the main unit

（On-site witnessing is required (for the system construction)）

By the network construction, can be sent to the network server the measurement results instantaneously.
Daily data accumulation is simple.Prevention of copy leakage.
Creating the form of measurement data from the office PC (Excel format). Possible to create a print data.

Wi-Fi
access point

Wi-Fi

data accumulation

LANdata management
daily
reportmonthlyreportnetwork server（NAS)

Office PC

Body weight (with monitor cover) about 12kg.

Body

Monitor cover is fixed with a magnet.

frontback

left

410

155

monitor cover386

USB
Power switch

Master switch

1040

Pedestal

AC100-240V

VGAOmega connector (K)RS232250

Display screen

CE modeTypespecification

outside dimensions

QCR-Win 2022 model
W410 × H360 × D155 mm (Except for the Monitor cover and Pedesta)

Display monitor

15 inch touch panel(supersonic type)

OS

Windows 10 IoT Enterprise 2019 LTSC

MemoryStorage memory
Measurement range
Interface
Power source
Weightaccessories

optionSG mode8GB
About 20GB
50 〜 1350℃ (JIS-K)
USB2.0 / RS232x2 / VGA / LAN
100-240V
12kg (body only)
・Cup stand (cable length: 5m)
・Pedestal・Monitor cover・Data extraction for media
・Touch panel pen

・LAN port conversion
・Printer・data communication

・Wired thermometer add
・TM-RW (Wireless thermometer communication)

Please contact us for more details

The corresponding
consumables

Please refer to the catalog of the shell cup.

NAKAYAMACo.,Ltd.
■ Haed Office 3-37-22 Kodama, Nishi-ku, Nagoya-City 451-0066 Japan
Tel:+81-52-521-1171 Fax:+81-52-521-1180
E-mail: info@nakayama-meps.co.jp

August, 2011 1st edition
May, 2022 6th edition■ Website; http://www.nakayama-meps.co.jp/■ Eastern Japan business officeTel: +81-24-545-6588 Fax: +81-24-544-6588