NAKAYAMA
Co.,Ltd.
Wireless immersion thermometer

HANDY-THERMO

Charge-type Wireless thermometer

TMW-2000-M
PrinterTM-PR
Wireless large display

TM-IW-M

Wireless receiver

TM-RW-M

HANDY-THERMO
Small and light handy type thermometer. Easy measurement ofmolten metal temperature.

特長Feature

Automatic peak hold function keeps maximum temperature during themeasurement .
Automatic power-off function (1 minite).
Indicator is large red LED.
Available for one week on a full charge (theoretical value).
Thermocouple type: B, S, R, K (standard: R) .
The specific small electricity that is a license-freeLine-of-sight distance: 20m(There is a condition difference)

System example

Use models: TMW-2000

A

Displayed on a large display unit

TM-IW-M

Use models: TM-IW-M

4.maximum temperature indication

B

Printer output
Use models: TM-PR, TM-RW-M
Print contentdate2016 / 2 / 10TM-PR

TM-RW-M

4.print

1. temperature measurement
2. maximum temperature detection
3. wireless data output

Ctime CH Temp
00:00 ch1 1450

Save data to a PC
Use models: TM-RW

TM-RWOffice PC

D

4.Temperature datacommunication output

5.Temperature data reception

Displayed on a large display unit +Save data to a PC
Use models: TM-IW-M-DO(Data output type)
TM-IW-M-DO
Office PC

4. maximum temperature indication
5. Temperature data communication output

6.Temperature datareception

CH type

Use models: TMW-2000-M
CH 2

CH 1

CH 0

Possible to up to three
Use models: TM-IW-M

CH 3
TM-IW-M

CH

Temp

0

Temperature of Device No.0

1

Temperature of Device No.1

2

Temperature of Device No.2

3

Temperature of Device No.3

*Note: Reception is an interval for three seconds.
The reception indication is impossible at the same time.

Multifrequency typeUse models: TMW-2000-M
C frequency

A frequency

A frequency

B frequency

B frequency

A frequency

C frequency

Output to each frequency band

Frequency band(A)Frequency band(B)

TM-PRTM-IW-M

TM-RW

Frequency band(C)TM-IW-M(Data output type)

Office PC

TypeDimensions

specification

Weight
Thermo sensor
Measurement range

TMW-2000-M
W110 × H180 × D200 mm
About 1kg
Type-R thermocouple ( JIS C 1602: 2015 )
50 ~ 1768 °C

Accuracy±(0.3% of rdg＋1°C)

Resolution

1°C (1°F)

Reference contact point compensation
Radio frequency±1°C (0 ~ 100°C)
Specific small power ( line-of-sight distance: 20m)
・Standard: 429MHz(Fixed)
・M-Type: 429.25, 429.35, 429.45, 429.65MHz (Select)

Indicator

Red LEDcharacter size: 14.2 ×10 mmsensor open or upper limit :batteries are low :Peak hold function

Holds the maximum data of increasing temperature.

Power off function

Switches off the power when increasing temperature is not.

Batteries
Recharging power source
Accessories
Typedisplay

Dedicated AC adapter：1setReception frequency

TM-IW-M

character

TM-IW-M-DO

Dimensions

W500×H150×D70mm

TypePrinterAC100 ~ 240V

Data output

WeightReceiverNickel hydrogen cell

Dimensions

TM-PR
W150×H150×D250mm
About 2kg

TypeTM-RW-M

Power source

AC100-240V

Printing method

Serial dot thermal printerPower source

Dedicated adapter

Reception frequency

Body*1：W130×H60×D220mm
Antennas* ：W65×H350×D85mm
2Weight

Red LED character size: 110 ×70 mm

About 3kg

WeightDimensions

429MHzInterface
Power source

429MHzRS232
AC100-240V

About 0.6kg*1 Including rear connector*2 Including Antennas

The corresponding consumables
Please refer to the catalog of the thermometer consumables.

NAKAYAMACo.,Ltd.

Jan. 2011 3rd edition
Jul. 2023 6th edition■ Haed Office; 3-37-22 Kodama, Nishi-ku, Nagoya-City 451-0066 Japan
Tel:+81-52-521-1171 / Fax:+81-52-521-1180
E-mail: nk-1972@nakayama-meps.co.jp■ Website; https://www.nakayama-meps.co.jp/■ Eastern Japan business office; Tel: +81-24-545-6588 / Fax: +81-24-544-6588