IECEx Certificateof Conformity
INTERNATIONAL ELECTROTECHNICAL COMMISSION
IEC Certification System for Explosive Atmospheres
for rules and details of the IECEx Scheme visit www.iecex.com

Certificate No.:

IECEx LCIE 21.0002X

Page 1 of 4Status:

Current

Issue No: 1

Date of Issue:

2023-01-11

Applicant:BÜRKERT SAS
Rue du Giessen
67220 TRIEMBACH AU VAL
FranceEquipment:

Flowmeter - Type: 8098 series

Optional accessory:
Type of Protection:

Ex ec and Ex tc

Marking:

Ex ec IIC T4 Gc
Ex tc IIIC T110°C Dc or T130°C Dc(Refer to attachment for full marking).

Approved for issue on behalf of the IECEx
Certification Body:

Julien GAUTHIER

Position:

Certification Officer

Signature:(for printed version)Date:(for printed version)

2023-01-11

1. This certificate and schedule may only be reproduced in full.
2. This certificate is not transferable and remains the property of the issuing body.
3. The Status and authenticity of this certificate may be verified by visiting www.iecex.com or use of this QR Code.

Certificate issued by:Laboratoire Central des Industries Electriques (LCIE)
33 Avenue du General Leclerc
FR-92260 Fontenay-aux-Roses
FranceCertificate history:
Issue 0 (2021-06-24)

IECEx Certificateof Conformity
Certificate No.:

IECEx LCIE 21.0002X

Page 2 of 4

Date of issue:

2023-01-11

Issue No: 1

Manufacturer:

Bürkert Werke GmbH & Co KG
Christian Bürkert-Straße 13-17
74653 Ingelfingen
Germany

Manufacturinglocations:BÜRKERT SAS
Rue du Giessen
67220 TRIEMBACH AU VAL
FranceThis certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with theIEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by this certificate, was assessed andfound to comply with the IECEx Quality system requirements.This certificate is granted subject to the conditions as set out in IECEx SchemeRules, IECEx 02 and Operational Documents as amended
STANDARDS :The equipment and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was foundto comply with the following standards
IEC 60079-0:2017
Edition:7.0

Explosive atmospheres - Part 0: Equipment - General requirements

IEC 60079-31:2013
Edition:2

Explosive atmospheres - Part 31: Equipment dust ignition protection by enclosure "t"

IEC 60079-7:2017
Edition:5.1

Explosive atmospheres - Part 7: Equipment protection by increased safety "e"This Certificate does not indicate compliance with safety and performance requirements
other than those expressly included in the Standards listed above.
TEST & ASSESSMENT REPORTS:
A sample(s) of the equipment listed has successfully met the examination and test requirements as recorded in:
Test Reports:
FR/LCIE/ExTR21.0004/00
Quality Assessment Report:
DE/PTB/QAR07.0002/11

FR/LCIE/ExTR22.0059/00

IECEx Certificateof Conformity
Certificate No.:

IECEx LCIE 21.0002X

Page 3 of 4

Date of issue:

2023-01-11

Issue No: 1

EQUIPMENT:
Equipment and systems covered by this Certificate are as follows:
The device is used to measure the flow of liquids.
It consists of a pipe in which a fluid circulates, an enclosure and a display.
(Refer to attachment for more details).
SPECIFIC CONDITIONS OF USE: YES as shown below:• The equipment shall only be used in an area of at least pollution degree 2, as defined in IEC 60664-1 standard.
• Transient protection shall be provided that is set at a level not exceeding 140% of the peak rated voltage value at the supply terminals tothe equipment.
• All entries in the enclosure shall be equipped with cable glands, connectors or blanking elements certified with a protection mode and an
IP compatible for the intended use.
• Any parallel threaded entry with less than five full threads engaged shall be used with an additional seal or gasket.
• Potential electrostatic charging hazard. Clean only with a wet cloth (see instructions).

IECEx Certificate
of Conformity
Certificate No.:

IECEx LCIE 21.0002X

Page 4 of 4Date of issue:

2023-01-11

Issue No: 1

DETAILS OF CERTIFICATE CHANGES (for issues 1 and above)
Issue 01:
Update to include pipe diameter up to DN 100/4"
Modification of the electronic board (remove relay function on ME39 board)
Update of the technical file accordingly

Annex:
Annex to Certificate IECEx LCIE 21.0002X - Issue 01.pdf

Annex to Certificate
IECEx LCIE 21.0002X issue 01
FULL EQUIPMENT DESCRIPTION
The device is used to measure the flow of liquids.
It consists of a pipe in which a fluid circulates, an enclosure and a display.
The flowmeter type 8098 series is a product of the FLOWave range. It uses the SAW (Surface Acoustic Waves) technology andis at first designed for the use in applications requiring that all hygienic conditions are fulfilled. The main use focus is on hygienic
applications and for the measurement or monitoring of water similar liquids.
As an example low conductivity or nonconductive water is a very preferred area of usage as FLOWave flow measurement isindependent from conductivity.
No parts in the measurement tube.
The housing enclosure is equipped with an already certified Ex component, see table below:
OperatingComponent Manufacturer
TypeCertificate
Ex marking
temperature
Protective
IECEx IBE
W. L. Gore &- 40°C
Vent type17.0013U
Ex eb IIC Gb
Associates
PMF200400
up toGore®
Date of Issue:
Ex tb IIIC Db
GmbH+125°C
PolyVent
2020-11-16

Applied standards
IEC 60079-0, Ed.7.0
IEC 60079-7, Ed.5.1
IEC 60079-31, Ed.2.0

MARKING

Address: …
Year of construction: …
Serial N°: …
Type: 8098*****
Ex ec IIC T4 Gc
Ex tc IIIC T110°C Dc or T130°C Dc
IECEx LCIE 21.0002X-10°C ≤ Tamb ≤ +40°C according to the process fluid temperature and the operation time.
Fluid temperature: -20°C … +130°C (See table below)
Ratings: 35V DC, 5W.
WARNINGS –DO NOT CONNECT OR DISCONNECT WHEN ENERGIZEDDO NOT OPEN WHEN ENERGIZEDPOTENTIAL ELECTROSTATIC CHARGING HAZARD - SEE INSTRUCTION (only in dust atmosphere)

Continuous operation allowed

Operation allowed for maximum 1 hour

Process fluidtemperature(°C)-20-100102030405060708090100110120130Ambient
temperature(°C)
40404040404040404040403836343230

Temperature class
No product orientation restriction
GasDustT4T110°C

T4T130°C

Page 1 of 3
This Annex is valid only in combination with certificate mentioned above and may only be reproduced in its entirety and without any change.
CERT-ATEX-FORM 14 Rev. 03

Annex to Certificate
IECEx LCIE 21.0002X issue 01

RANGE DETAILS
(1)
TYPE

(2)
DN

(10)
DG

(12)
TTYP

(13)
ELEA

Description
(1)
TYPE

(2)
DN

(10)
DG
(12)
TTYP
(13)
ELEA
(14)
ELEW
(15)
ERM1(16)
ERM2(17)
ERM3(18)
ERM4(24)
IO(31)
IECProduct type

Orifice size

Display size

(14)
ELEW

(15)
ERM1(16)
ERM2

Product range
Allowed value
8098
8.0
10.0
15.0
20.0
25.0
40.0
50.0
65.0
80.0
100.0
⅜”-½”-¾”-1”-1½”-2”-2 ½”-3”-4”
01(17)
ERM3(18)
ERM4(24)
IO

(31)
IEC

Signification
FLOWave flowmeter

Measurement pipe diameter

0: no display
1: 2.4” display

Transmitter type02

SE98

Electrical connection

RB2 cable glands, 1xM12

Electrical connection
material

2

Stainless steel

Extension module 10

Not applied for an Ex usingExtension module 20

Not applied for an Ex using

Extension module 30

Not applied for an Ex using

Extension module 40

Not applied for an Ex using

Input / Outputs

S1 AO + 1 DO + 1 AO/DOIECEx approval

0001

IECEx 3GD

RATINGS
U:35V DC, P:5W, with electronic charge 700mA

ROUTINE TESTS
None.

Page 2 of 3
This Annex is valid only in combination with certificate mentioned above and may only be reproduced in its entirety and without any change.
CERT-ATEX-FORM 14 Rev. 03

Annex to Certificate
IECEx LCIE 21.0002X issue 01APPARATUS OVERVIEW: example

Page 3 of 3
This Annex is valid only in combination with certificate mentioned above and may only be reproduced in its entirety and without any change.
CERT-ATEX-FORM 14 Rev. 03