FJEABNRUUAARRYY

2 0 2 43

OR T HOPEDIC INS T RUMEN T S

Yezerski Small Bone Rongeurs
Page 3

Faillace Extra
Small Bone Clamp
Page 15

Featuring
many

instruments
throughout

AndersonTalar Neck Osteotome
Page 27

Joint, Calcaneal, and Small Bone
Compressor/Distractors with
Speed Lock
Page 13

Johnson Low Profile
Foot & Ankle Retractors
Page 5

Foot & Ankle Instruments
1.800.548.2362   INNOMED.NET®

Fromm Femur & Tibia Triangles
Designed by S.E. Fromm, MD *Extra Small Triangle designed by S.E. Fromm, MD & Kenneth Merriman, MD

Used for femur and tibia positioning
during nailing, repairs and fractures
Designed to position and hold the femur and tibia during
intramedullary nailing of the tibia, ligament repairs andextremity fractures. Allows knee to be flexed greater than 90° toallow reaming and nail insertion without displacing fracture. The
triangles are available in four heights: 8.5", 11", 14", and 16".
The three smaller triangles are designed to fit inside the larger
triangle for storage. They are supplied with an autoclavablesilicone cushioning pad and velcro* straps. The triangles arealso radiolucent and gas or steam sterilizable.
PRODUCT NO’S:

2760-00 [Set of 3] Angles: Top 30°, Two Bottom 75°

16” #2760-03

14” #2760-02

11” #2760-01

Set Includes / Available Individually:

Separately –8.5” #2760-XS Sold
Not In Set

2760-01 [11"] Base: 6" (15,2 cm), Height: 11" (27,9 cm)
2760-02 [14"] Base: 7" (17,8 cm), Height: 14" (35,6 cm)
2760-03 [16"] Base: 9" (22,9 cm), Height: 16" (40,7 cm)
Sold Separately – Not In Set:

2760-XS [8.5"] Base 5" (12,7 cm), Height: 8.5" (21,6 cm)
Replacement Parts:

2760-P [Silicone Pad]
2760-S [Straps] Package of 18
8120-SP [Straps for XS] Package of 10

Tibia Reduced For:

*Velcro is a registered trademark of the Velcro Companies.
®

• Open Reduction and Internal Fixation (ORIF)• Application of uni- or multi-plane external
fixator• Knee ligament repairs and/or reconstruction

RetrogradeFemoral Nailing
Triangle holds femur reduced(prevents sagging)

RetrogradeFemoral Nailing

Tibial Nailing

Sanders Extremity Positioning Tubes
Designed by Richard A. Sanders, MD

Designed to support the knee and ankle duringlower extremity surgery
The 6" tube lifts the knee off the operating table and allows for approximately 30°
of knee flexion. Very useful for closure of total knee incisions, supporting fracturesof the distal femur, and tibia plateau fractures. The 4" tube elevates the foot andankle for ankle fracture surgery. The tubes are made of aluminum, allowing them
to be autoclaved. They help eliminate the need for rolled sheet bolsters.
PRODUCT NO’S:

2740-01 [Small]4" #2740-01

6" #2740-02

Diameter: 4" (10,2 cm)
Width: 8" (20,3 cm)

2740-02 [Large]

Diameter: 6" (15,2 cm)
Width: 8" (20,3 cm)

Lower Extremity
Leg Positioner
Designed by Ronald Romanelli, MD

Used to support knee and leg during
surgery, and can be used for casting
Utilized for rodding of femurs or tibias
Also useful for knee surgery and closures
Very supportive, distributes stresses on leg,
used instead of bolsters
Supplied with one autoclavable silicone pad
Aluminum positioner is radiolucent and gas
or steam sterilizable

PRODUCT NO’S:

2745

Dimensions: 5.5" H x 9.5" L x 9.25" W      (12,7 cm x 24,1 cm x 23,5 cm)

Replacement Parts:

2760-P [Silicone Pad]

2

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Mazzara Rongeur for Small Bones
Designed by James T. Mazzara, MD

Designed for bone and soft tissue removal
in small joint surgery, the pistol grip handle
lessens hand fatigue and slippage, andallows for better visualization

2 x 10 mm Jaw Bite
#1765-04

4 x 10 mm Jaw Bite
#1765-05

PRODUCT NO’S:

1765-04

Jaw Bite: 2 x 10 mm
Overall Length: 9" (22,9 cm)

1765-05

Jaw Bite: 4 x 10 mm
Overall Length: 9" (22,9 cm)

2 x 10 mm

4 x 10 mm

Two Jaw Sizes Available

Yezerski Small Bone Rongeurs
Designed by John Yezerski, MD

Designed for small bone
applications in the hand and footPRODUCT NO’S:

Small #1789

1789 [Small]

Overall Length: 7.125" (18,1 cm)
Jaw Width: 4 mm
Jaw Bite Width: 3 mm
Jaw Bite Length: 20 mm

1789-01 [Extra Small]

Overall Length: 4.5" (11,4 cm)
Jaw Width: Tapers from 4,6 mm to 2 mm
Jaw Bite Length: 11 mm

Extra Small #1789-01

Macko
Square Tipped
Rongeur

7 x 18 mm Jaw Bite #1778-02
10 x 18 mm Jaw Bite #1778-03

Designed by Victor W. Macko, MD
PRODUCT NO’S:

1778-02

Jaw Bite: 7 x 18 mm
Overall Length: 10" (25,4 cm)

1778-03

Jaw Bite: 10 x 18 mm
Overall Length: 10" (25,4 cm)

Unique square tipped rongeur features anergonomic grip, double action mechanism,
long reach, and low profile for use in total
ankle, knee, hip, and spine surgery
When used for morcelizing bone graft, the shallow,
wide jaw helps avoid impaction.

PRODUCT NO:

1148

Overall Length: 4.75" (12,1 cm)
Large End Blade Length: 1.75" 4,4 cm)
Large End Blade Width: .625" (1,6 cm)
Small End Blade Length: 1" (2.5 cm)
Small End Blade Width: .3125" (0,8 cm)

1.800.548.2362       FEBRUARY 2024   

Kawell Short
Army Navy Retractor

Designed by Ron Kane, DPM

A short handled
Army Navy retractor,
especially useful with agastrocnemius recession    FOOT & ANKLE INSTRUMENTS

3

PRODUCT NO’S:

1159 [Standard Sharp Rake]
Overall Length: 4.5" (11,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

Chung T-Handle Retractors

Rake

Designed by Raymond Chung, MD

1161 [Standard Blunt Rake]
Overall Length: 4.5" (11,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

1162 [Standard Senn]

Designed with a T-handle for easier holding
and to help reduce finger and thumb fatigue
Standard Shaft

Overall Length: 4.5" (11,4 cm)
Blade Width: 6 mm
Blade Depth: 16 mm

Sharp Rake
Standard Shaft

Sharp Rake
Extended Shaft#1159-01

Extended Shaft#1159

1159-01 [Extended Sharp Rake]
Overall Length: 5.625" (14,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

Blunt Rake
Extended Shaft

Blunt Rake
Standard Shaft

1161-01 [Extended Blunt Rake]#1161-01#1161

Overall Length: 5.625" (14,4 cm)
Blade Width: 9 mm
Blade Depth: 7 mm

Senn
Extended Shaft

Senn
Standard Shaft

1162-01 [Extended Senn]#1162-01#1162

Overall Length: 5.625" (14,4 cm)
Blade Width: 6 mm
Blade Depth: 16 mm

Senn

Modified
Mini Hohmann
Retractors

Superior Coracoid
Modification

PRODUCT NO’S:

1665 [Blade: 6 mm Wide / 35 mm Drop]
Overall Length: 5.875" (14,9 cm)
Blade Width: 6 mm
Blade Drop: 35 mm

Designed by Jeffrey Lawton, MD

1665-01 [Blade: 6 mm Wide / 17 mm Drop]

Used for smallbone surgery

Overall Length: 5.5" (14 cm)
Blade Width: 6 mm
Blade Drop: 17 mm

1666 [Blade: 8 mm Wide / 35 mm Drop]
Overall Length: 5.875" (14,9 cm)
Blade Width: 8 mm
Blade Drop: 35 mm

1666-01 [Blade: 8 mm Wide / 17 mm Drop]

6 mm Wide /35 mm Drop
#1665

Overall Length: 5.5" (14 cm)
Blade Width: 8 mm
Blade Drop: 17 mm

6 mm Wide /17 mm Drop
#1665-01

8 mm Wide /35 mm Drop#1666

8 mm Wide /17 mm Drop
#1666-01

1666-02 [Blade: 8 mm Wide / 17 mm Drop]

8 mm Wide / 17 mm Dropwith Superior Coracoid
Modification

7 mm Wide / 72 mm Drop
#1666-LG

Overall Length: 6.25" (15,9 cm)
Blade Width: 8 mm
Blade Drop: 17 mm

1666-LG [Blade: 8 mm Wide / 72 mm Drop]
Overall Length: 7.125" (18,1 cm)
Blade Width: 8 mm
Blade Drop: 72 mm#1666-02

11 mm #1643-11

13 mm #1643-13

McGlamry Type Elevators
Designed to help deglove ametatarsal head, and helpful
in many other procedures
PRODUCT NO’S:

1643-11 [11 mm]

Overall Length: 6.5" (16,5 cm)

15 mm #1643-15

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1643-13 [13 mm]

Overall Length: 6.5" (16,5 cm)

1643-15 [15 mm]

Overall Length: 6.5" (16,5 cm)

1643-17 [17 mm]

17 mm #1643-17

4

WWW.INNOMED.NET   Overall Length: 6.5" (16,5 cm)   FREE TRIAL ON MOST INSTRUMENTS

Johnson Low Profile
Foot & Ankle Retractors
Designed by Michael Johnson, MD

Designed for soft tissue
retraction in the foot and ankle

PRODUCT NO’S:

1636-01 [Straight]

Bent

Overall Length: 7.5" (19 cm)
Blade Width: .41" (1 cm)

Straight#1636-02

1636-02 [Bent]#1636-01

Overall Length: 8.875" (22,5 cm)
Blade Width: .41" (1 cm)

OrthoLucent™ Mini Hohmann Retractors
Designed by Jeffrey Lawton, MD

Radiolucent, lightweight retractors
The carbon fiber PEEK material is strong, lightweight,
completely radiolucent, can be steam sterilized, and helps
to prevent from marring component surfaces.

8 mm Blade #1594-R

PRODUCT NO’S:

1594-R [8 mm Blade]

Overall Length: 6.875" (17,5 cm)
Blade Width: 8 mm

1597-R [16 mm Blade]
Overall Length: 6.875" (17,5 cm)
Blade Width: 16 mm

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

16 mm Blade #1597-R

Swanson Elevator
Designed by Richard Ferkel, MD

Angular design helps to goaround bone for retractionand elevation — especially
useful in small bone
surgery of the foot/ankle
and hand/wrist
PRODUCT NO:

1644

Overall Length: 6.375" (16,2 cm)
Blade Depth: .75" (1,9 cm)

J.B. Redler Retractor
Designed by M.R. Redler, MD
PRODUCT NO:

1645

Overall Length: 5" (12,7 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       FEBRUARY 2024   

Uniquely balanced retractor for bone exposure
for a multitude of upper extremity procedures
Double-angle design allows for ideal exposure with minimal effort to hold theretractor, while the assistant's hands are well out of the way of the exposure.
The aperture in the base of the handle allows the retractor to be attached via
a Penrose drain to the table for hands-free approach.

    FOOT & ANKLE INSTRUMENTS

5

Faillace Ambidextrous
Self-Retaining Retractor
Designed by John J. Faillace, MD

Handle can be rotated away fromthe surgeon after insertion if desired
PRODUCT NO’S:

1580 [7 Teeth]

Overall Length: 7.5" (19,1 cm)
Prong Depth: 38 mm
Prong Width: 34 mm

7 Teeth#1580

1579 [4 Teeth]

Overall Length: 6" (15,2 cm)
Prong Depth: 38 mm
Prong Width: 18 mm

1579-01 [Small – 4x3 Teeth]

4 TeethOverall Length: 5.25" (13,3 cm)
Prong Depth: 20 mm
Prong Width: 18 mm / 13 mm#1579

4x3 Teeth#1579-01

Dodson Modular Retractor

Designed to help expose a small to medium sizebone for internal fixation—can be used for distal
radius, ulna, humerus, and fibula fractures

Designed by Mark A. Dodson, MD

Allows the limb to be rotated (pronated or supinated)without loss of exposure. The hohmann retractors have threehole sizes which allow for a variety of positioning angle
options using the teeth of the self-retaining retractor, or canalso be positioned in-between the teeth. The hohmann isplaced around the bone, and thus reduces the force on thesoft tissues while increasing exposure. Can be used in theforearm to treat radius and ulna shaft fractures, humerus
fractures, as well as in the leg for fibula fractures.
PRODUCT NO’S:

Set consists of
one ratcheting selfSet Includes / Available Individually:retaining retractor,
1838-01 [Retractor Only]
two stainless steel
Overall Length: 5.5" (14cm)
mini-hohmann
1838-02 [Blade Only – One] retractor blades, anda stertilization case.
Overall Length: 5.25" (13,3cm)
1838-00 [Set]

Blade Width: 3/8" (9mm)1025 [Sterilization Case Only]
Optional Parts — Not Included In Set:

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY* MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLANDUS Patent No. 9,161,745 B2

1838-02R* [Radiolucent Blade Only – One]
Overall Length: 5.25" (13,3cm)
Blade Width: 3/8" (9mm)

Optional radiolucent carbon fiber PEEK composite blade
The optional radiolucent blade is made of a strong, lightwieght carbon fiber
PEEK composite material, which is completely radiolucent, helps to prevent frommarring component surfaces, and can be steam sterilized.

Hendren Neuroma Retractor
Designed by Douglas H. Hendren, MD

Narrow tines are delicate on tissue,
but sturdy enough to retract bone
Provides excellent exposure. Also helpfulin scaphoid fracture repair surgery.
PRODUCT NO’S:

1680-01 [Small]

Overall Length: 4.25" (10,8 cm)

1680-02 [Large]

Overall Length: 5.5" (14 cm)

6

Large

#1680-02

WWW.INNOMED.NET   

Small#1680-01   FREE TRIAL ON MOST INSTRUMENTS

Wurapa Swivel Blade Retractor
Designed by Raymond Wurapa, MD

Designed for forearm and wrist fracture exposure,the blades swivel for less stress on soft tissue
Swivel-blade technology helps to allow parallel deployment of retractor blades
to maximize wound exposure and minimize edge loading on surrounding
soft tissues. Parallel deployment of the retractor blades also helps prevent
rotation and migration of the retractor during a procedure.
PRODUCT NO’S:

1646-00 [Set]

Includes Retractor and Two Swivel Blades

Set Includes / Available Individually:

1646-01 [Retractor]

Overall Length: 5.125" (13 cm)
Opens to: 2.5" (6,4 cm)

1646-02 [Swivel Blade]

One blade with this product number, two included in set

Width: .9375" (24 mm)
Depth: .75" (19 mm)

Chung Weitlaner Retractors
Designed by Raymond Chung, MD

Longer prongs allow use ina small, but deep wound

Prong lengths of 25 mm and 30 mm
available with either sharp or blunt tips

25 mm

25 mm 3x4 Blunt Prongs #5065-01
25 mm 3x4 Sharp Prongs #5066-01
30 mm 3x4 Blunt Prongs #5067-01
30 mm 3x4 Sharp Prongs #5068-01
PRODUCT NO’S:
3x4 Prongs — Blunt Tips

3x4 Prongs — Sharp Tips

PRODUCT NO’S:
2x3 Prongs — Blunt Tips

2x3 Prongs — Sharp Tips

5065-01 [25 mm]

5066-01 [25 mm]

5065 [25 mm]

5066 [25 mm]

5067-01 [30 mm]

5068-01 [30 mm]

5067 [30 mm]

5068 [30 mm]

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)

30 mm

25 mm 2x3 Blunt Prongs #5065
25 mm 2x3 Sharp Prongs #5066
30 mm 2x3 Blunt Prongs #5067
30 mm 2x3 Sharp Prongs #5068

Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Blade Depth: 25 mm
Overall Length: 4.5" (11,4 cm)
Blade Depth: 30 mm
Overall Length: 4.5" (11,4 cm)

Monaco Small Space Retractor
Designed modified by Spencer Monaco, DPM, FACFAS

Designed to retract adipose
tissue and surrounding soft tissue
structures through a small incision
for open plantar fasciotomies,
neuroma excisions and the lateralrelease during bunion surgery
Also useful for various hand surgeries
such as open carpal tunnel surgery.
PRODUCT NO:

1887-01

Overall Length: 4.25" (10,8 cm)
Blade Depth: 18 mm
Blade Width: 12 mm
Blade Lip: 3.5 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTS

7

Ortho Self-Retaining
Retractor with Pin Guides
Designed by Sean Dunn, DPM

Designed to distract a small joint during
fusion or osteotomy alignment surgery
PRODUCT NO:

1842-02

Overall Length: 6.5" (16,5 cm)
Blade Width: 7 mm
Blade Extension (beyond guides): .4" (1 cm)
Blade Thickness: 1.68 mm
Pin Guide Length: 1.25" (3,2 cm)
Pin Guide Internal Diameter: .085" (2,1 mm)

Calibrated
Ortho Spreader
with Slotted Tips

Locations for useof Calibrated Ortho
Spreader with
Slotted Tips

Designed by Jason Bariteau, MD

A lamina spreader witha very thin closed profile,designed to enabledistraction in tightspaces like the subtalar
and talonavicular joints

PRODUCT NO:

1841

Overall Length: 6.75" (17,1 cm)
Prong Length: .5" (12,7 mm)
Calibrations: 10 mm to 35 mm

Calibrations on Handle

HFD Self-Retaining
Small Bone Spreader

Versatile spreader featuring narrow
tapered blades which, when together,make a small wedge to enter a tightbone interface or osteotomy
Blades feature a non-aggressive grip pattern thatcan be used when spreading apart bone as well as
providing retraction of soft tissue in a smaller wound.
PRODUCT NO:

1829

Overall Length: 4.5" (11,4 cm)
Blade Depth: 28 mm
Blade Width Tapers from: 8 mm to 5 mm

Calcaneal Lateral
Column Spreader
Designed by K. Wapner, MD

For lateral column
lengthening of the calcaneus
PRODUCT NO:

1725

Pads: 14 mm x 12 mm
Arms Open to: 45 mm
Overall Length: 4.25" (10,8 cm)

8MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Weinraub Joint and Calcaneal Spreader

Speed lock helps allow
precise control and prevent
unintended release

Designed by Glenn M. Weinraub DPM, FACFAS

Designed to assist in the opening of
small joints of the foot and hand for theapplication of fusion and graft techniques

ORIGINAL
DESIGN!

Provides excellent joint exposure without blocking intraarticularor osteotomy access. Helps prevent slippage or falling out of
the joint by placing the arms on either side of the area to be
distracted, driving two pins and opening the joint.
PRODUCT NO’S:

1870 [Standard 1.6 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .062" (1/16") (1.6 mm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY1872 [Standard 2.8 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .11" (7/64") (2.8 mm)

1870-SL [Speed Lock 1.6 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .062" (1/16") (1.6 mm)

1872-SL [Speed Lock 2.8 mm]

Overall Length: 7" (17,8 cm)
Pin Diameter: Up to .11" (7/64") (2.8 mm)

Calcaneal Spreader

1.6 mm Standard #1870
2.8 mm Standard #1872

1.6 mm Speed Lock #1870-SL
2.8 mm Speed Lock #1872-SL

Designed by Michael Forness, DO

Separates the calcaneal osteotomized bone for placement of tricortical bone graft
Pads have a large surface area, which easily separates the calcaneal osteotomized bone for placement of tricorticalbone graft. Large pad surface area helps prevent the compression of soft calcaneal cancellous bone.

Smooth Pads
Smooth Pads #1880

PRODUCT NO’S:

1880 [Smooth Pads]

Overall Length: 7" (17,8 cm)
Pad Dimensions: 15 mm x 12 mm

1881 [Grooved Pads]

Grooved Pads

Grooved Pads #1881

Overall Length: 7" (17,8 cm)
Pad Dimensions: 15 mm x 12 mm

Strayer Retractor
Designed by Irvin Oh, MD

A lamina spreader
with long thin blades
designed to retract
the soleus muscle andsoft tissue for isolationand exposure of thegastrocnemius fascia
for release
PRODUCT NO:

1869

Overall Length: 9.25" (23,5 cm)
Blade Length: 3.5" (8,9 cm)
Blade Width: .6" (1,5 cm)

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTS

9Desai Clearview Open BladeSelf-Retaining Retractor
Designed by Sarang Desai, DO

Open blade design allows clear
visualization of soft tissue andneurovascular structures being
retracted
Tapered blades allows 90° deep soft
tissue retraction and easy insertion into
the wound. The open blades also allow
surgeon to work in open blade area, suchas for gastroc recession surgery.

PRODUCT NO:

1858

Overall Length: 7.25" (18,4 cm)
Blade Depth: 3" (7,6 cm)
Blade Width: 1.25" (3,2 cm)

Percutaneous Achilles Repair Forceps
FOR LIMITED OPEN ACHILLES TENDON REPAIR
Designed by James A. Amis, MD

Designed to help improve accuracy duringpercutaneous repair of Achilles tendon ruptures
Lateral Bump

The bump on the lateral side ofeach loop allows the surgeon
to palpate the exact center ofthe loop, proximal to distal,and drop a needle just below(patient is prone) or anterior to
the bump for the starting point,
and aim to just below thebump on the opposite side.

PRODUCT NO:

8235

Overall Length: 9.625" (24,4 cm)

Gurbani Joint Distractor/Compressor
Designed by Naren G. Gurbani, MD

Versatile joint distractor/compressor for arthroscopic
or open procedures of foot, ankle, hand, and wrist joints
The surgeon puts the pins in the bone, then slides the holes of the device overthe pins and distracts or compresses—the device can be locked in either direction.
Especially useful for arthroscopy of subtalar, talo-navicular, calcaneo-cuboid, andwrist joints. The T-wrench helps provide precise, controlled manipulation.

Pin Hole Sizes:.15" (3,5 mm) and.182" (4,5 mm)

PRODUCT NO’S:

4208-00 [Set]

Includes: Distractor/Compressor, T-Wrench, and Case

Set Includes / Available Individually:4208-01 [Distractor/Compressor Only]

Dimensions: 6" w x 5" h (15,2 cm x 12,7 cm)
Distracts up to: 3" (7,6 cm) / Compresses down to: .5" (1,3 cm)

4208-TW [T-Wrench]

Dimensions: 3" w x 3" h (7,6 cm x 7,6 cm)

1025 [Sterilization Case]

10

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Mantis Screwdriver Distractor
Designed by J. Albert Diaz, MD

Designed to help provide stable
distraction across difficult-to-reduce
fractures using two seated screwdrivers**Screwdrivers not included.
PRODUCT NO:

3654

Overall Length: 7.5" (19,1 cm)
Pin Hole Diameters: 4.5, 5.5, & 8.5 mm
Leg & Pin Hole Depth: .7" (17,5 mm)

Accommodates screwdrivers of varying sizefor use with both small and large fragment systems
Allows for distraction of difficult-to-reduce fractures
without the need to drill additional holes outside of the plate
The plate can be locked with a screw once length has been restored

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYHFD Compressor/Distractor

PRODUCT NO’S:
SMALL

1834 [Small – All Stainless Steel]

Dial mechanism helps allow precise
control of inserted wires— for maintaining
a position, compressing or distracting

Dimensions: 51 mm x 57 mm
Maximum Arm Opening: 1.35" (3,4 cm)

1834-R [Small w/Radiolucent Arms]
Dimensions: 51 mm x 57 mm
Maximum Arm Opening: 1.35" (3,4 cm)

A .125" (3,2 mm) pin can be used in theholes of the thumbwheel for leverage
Small: Two hole sizes allow for ease of pin size
selection: .045" (1,1 mm) & .062" (1,6 mm)
Large: Two hole sizes allow for ease of pin size
selection: .082" (2,0 mm) & .125" (3,2 mm)
Radiolucent arms are a steam sterilizable
PEEK/Carbon Fiber composite

LARGE

1836 [Large – All Stainless Steel]

Large with StainlessSteel Arms #1836

Overall Length: 4" (10,2 cm)
Maximum Arm Opening: 2.25" (5,7 cm)

1836-R [Large w/Radiolucent Arms]
Overall Length: 4" (10,2 cm)
Maximum Arm Opening: 2.25" (5,7 cm)

Small with StainlessSteel Arms #1834
Large with
Radiolucent Arms
#1836-R

Small with
Radiolucent Arms
#1834-R

Wurapa Small Joint
Compressor and Distractor

Available with two holesizes on each instrument

Designed by Raymond K. Wurapa, MD

Designed to allow one-handed
manipulation and deployment
once fixation pins are placed
K-wires should be cut short above the pin
guides to allow full access to the operative site

COMPRESSOR
1.1 & 1.6 mm Holes #1751
Single 1.1 mm Hole #1753

1.800.548.2362       FEBRUARY 2024   

1751 [Compressor]

Compresses From: 28 mm
Overall Length: 4.625" (11,7 cm)

Designed to simplify several
small joint procedures:
Preparation of small bone non-unions
before bone grafting and fixation
Preparation of small joints for arthrodesis(e.g. partial wrist fusion)
Distract and better evaluate small joints
before determining final management
Useful for intercarpal stabilization while
performing ligament reconstructions(e.g. scapholunate ligament repair/reconstruction)

PRODUCT NO’S:DOUBLE HOLES:.045" (1,1 mm) & .062" (1,6 mm)

1752* [Distractor]

Distracts to: 46 mm
Overall Length: 4.625" (11,7 cm)

SINGLE HOLE:.045" (1,1 mm) Hole

1753 [Compressor]

Compresses From: 28 mm
Overall Length: 4.5" (11,4 cm)

DISTRACTOR
1.1 & 1.6 mm Holes #1752
Single 1.1 mm Hole #1754

1754 [Distractor]

Distracts to: 46 mm
Overall Length: 4.5" (11,4 cm)

MADE EXCLUSIVELYFOR INNOMED IN*

GERMANY    FOOT & ANKLE INSTRUMENTS

11

Joint, Calcaneal andSmall Bone Distractors

Two hole sizes and two arm designs allow
for easier pin size selection and helps with
distraction in a variety of indications
PRODUCT NO’S:OUTSPREAD ARMS

4210-LB [Large]

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

4210-SB [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Large Outspread Arms
#4210-LB

CLOSED ARMS

4210-LS [Large]

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

4210-SS [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

4210-XSD [Extra Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 4.25" (10,8 cm)

Large Closed Arms

Two hole sizes allow for pin size selection:.062" & .094" / 1,6 mm & 2,4 mm#4210-LS

Extra Small#4210-XSD

Small Outspread Arms
#4210-SB

Small Closed Arms
#4210-SS

Joint, Calcaneal and Small BoneDistractors with Thumbscrews

Two hole sizes allow for pin size selection:.062" & .094" / 1,6 mm & 2,4 mm

Thumbscrew Modification Designed by Kelly McCormick, MD

PRODUCT NO’S:OUTSPREAD ARMS

CLOSED ARMS4215-LB [Large]

4215-LS [Large]

4215-SB [Small]

4215-SS [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)
Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Thumbscrews help prevent theunit from sliding on the pins

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

Large Outspread Arms with Thumbscrews #4215-LB
Small Outspread Arms with Thumbscrews #4215-SB
Large Closed Arms with Thumbscrews #4215-LS
Small Closed Arms with Thumbscrews (Shown) #4215-SS

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Joint, Calcaneal, andSmall Bone Compressor

Designed for compression in
fracture and osteotomy procedures

Small

Two hole sizes for ease of pin size selection:.062" (1,6 mm) & .094" (2,4 mm)#4210-SCPRODUCT NO’S:

4210-SC [Small]

Overall Length: 6" (15,2 cm)

4210-XSC [Extra Small]

Extra Small#4210-XSC

Overall Length: 4.25" (10,8 cm)

12

WWW.INNOMED.NET   

Two hole sizes allow for pin size selection:.062" & .094" / 1,6 mm & 2,4 mm   FREE TRIAL ON MOST INSTRUMENTS

Joint, Calcaneal, and Small Bone
Compressor/Distractors with Speed Lock
Speed lock helps allow precise control
and prevents unintended release

Thumbscrew version locks
the unit on the pins to help
prevent up or down sliding
Large Closed Arms
with Speed Lock
#4216-LS

Small Closed Arms
with Speed Lock
#4216-SS

Large Outspread
Arms with Speed
Lock & Thumbscrews

PRODUCT NO’S:
CLOSED ARMS WITH SPEED LOCK

4216-LS [Large]

OUTSPREAD ARMS WITH
SPEED LOCK & THUMBSCREWS

Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
4217-LB [Large]
Overall Length: 8" (20,3 cm)
Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
4216-SS [Small]
Overall Length: 8" (20,3 cm)
Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
CLOSED ARMS WITH
Overall Length: 6" (15,2 cm)
SPEED LOCK & THUMBSCREWS

4216-XS [Extra Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 4.5" (11,4 cm)

Extra Small Closed
Arms with Speed Lock

Small Closed Arms with
Speed Lock & Thumbscrews#4217-LB#4217-SS#4216-XS

Two hole sizes allow for pin size selection:.062" & .094" / 1,6 mm & 2,4 mm

4217-SS [Small]Holes Diameters: For .062" & .094"(1,6 & 2,4 mm) K-wire Pins
Overall Length: 6" (15,2 cm)

Large Pin
Distractor and Compressor

Larger 1/8" (3,2 mm) pin hole size forextra sturdy distraction or compression

Compressor#4234

PRODUCT NO’S:

4233 [Large Pin Distractor]

Hole Diameters: For .125" (3,2 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

4234 [Large Pin Compressor]

Hole Diameters: For .125" (3,2 mm) K-wire Pins
Overall Length: 8" (20,3 cm)

Distractor#4233

Joint, Calcaneal, and Small
Bone Compressor/Distractor

Selection Lever

Selection lever switches the mechanism
from compression to distraction
Simply squeeze the handle one time after
direction selection to engage the mechanism.
Two hole sizes for pin size selection.
PRODUCT NO:

4865-LS

Overall Length: 8.5" (21,6 cm)
Holes For: .062" & .094" (1,6 & 2,4 mm) K-wire Pins

1.800.548.2362       FEBRUARY 2024   

Two hole sizes allow for pin size selection:.062" & .094" / 1,6 mm & 2,4 mm    FOOT & ANKLE INSTRUMENTS

13

Wixted Fracture Distractor
Designed by John J. Wixted, MD

Designed to provide opposing leverageto help bring the fibula (or other bone)back out to its proper length after it
has been shortened by a fracture
PRODUCT NO:

1882

Overall Length: 7" (17,8 cm)

A 3.5 mm screw is temporarily placed above
a plate, providing a source of leverage forthe screw holding end of the distractor. The
curved peg-shaped tip is then placed intoa hole in the bone plate, and the distractor
is activated to bring the bone back to its
proper length before fixation.

Curved Peg-shaped Tip

Cut-out for Screw

Provides a secure source ofleverage against a temporarily
placed 3.5 mm screw

Fits securely into a hole in abone plate for leverage

Ratcheting Reduction Clamp Kit
Designed by Michael Craig, OPA-C

Designed as a soft tissue sparing
fracture reduction clamp
PRODUCT NO’S:

3840-00 [Clamp Kit]
Kit Includes / Available Individually:

3840-02 [Plate Point]

Overall Length: 1" (2,54 cm)

3840-03 [Screw Point]

Overall Length: .875" (2,2 cm)

3840-04 [Percutaneous Point]

2 included in set, one with this product number
Overall Length: 1" (2,54 cm)

Plate
Point

3840-MA [Ratcheting Reduction
Mobile Arm with Ratchet Knob]

Screw
Point

Percutaneus
PointsOverall Length: 6.5" (16,5 cm)

3840-SA [Ratcheting Reduction
Stationary Arm]
Overall Length: 10.5" (26,7 cm)
Width: 9" (22,9 cm)
Height: 6" (15,2 cm)

Kit includes:(1) Ratcheting Reduction Stationary Arm,
(1) Ratcheting Reduction Mobile Arm with Ratchet Knob,(1) Plate Point, (1) Screw Point, and (2) Percutaneous Points

High torque can help provide bone and joint
reduction without squeezing surrounding tissues
Swivel points are placed on the bone, plate, or
screw and the ratcheting dial is turned to thedesired torque, allowing hands free operation
Swivel point design allows the clamp to be easily
moved from x-ray view without losing reduction
Screw Point fits into a screw head
Plate Point fits into a 3.5 mm plate hole

PRODUCT NO:

3664

Medial Malleolus
Fracture Reduction Aid

Overall Length: 7" (17,8 cm)
Handle Length: 4" (10,2 cm)
Plate Width: .8" (2 cm)
Plate Length: 3" (7,6 cm)
Guide Tube Length: 6 mm

Designed by Christopher Blair, DO

Designed to hook under the medial malleolus
to help reduce the medial malleolus fragment
while two K-wire guides supply trajectory for wires
For K-wires up to 1,6 mm (.062")

14

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Bush Small Bone Reduction Forceps
Designed by Andrew P. Bush, MD

Designed to help hold a small bone or boneplate in position for reduction and fixation

PRODUCT NO’S:

1889 [Single]

Overall Length: 4.5” (11,4 cm)
Jaw Width: .15” (3,7 mm)

1888 [Double]

Overall Length: 4.5” (11,4 cm)
Jaw Width: .7” (17,7 mm)

Opens to approximately .5” (13 mm).

SingleDouble

#1889

#1888

Durham Bone Reduction Clamp
Designed by Alfred A. Durham, MD

Allows application of a bone plate
without removing the reduction clamp—
designed for medium size bones suchas the fibula, ulna, and radius

Large Window

Provides space to allow a boneplate to be slid into position
without removing the clamp

PRODUCT NO:

3652

Overall Length: 7" (17,8 cm)

Bargo Bone Holding Clamp

Designed by Lonnie Bargo, CST/CFA

Designed to aid in the reduction of various fractures,and can help secure a plate in place during installation
Designed to aid in the reduction of various fractures such as: spiral, transverse,
compound, oblique, or butterfly. The clamp can also be used to secure a plate in place
while the screw holes are being drilled and screws inserted. The fracture site can also
be manipulated with the clamp being used as a lever. Teeth in the jaws allows for abetter grip and a ratchet locking handle allows use on various bone diameters.
PRODUCT NO:

1803

Cannula Diameter: .062" (1.6 mm)
Overall Length: 5.25" (13,3 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Faillace Extra Small Bone Clamp
Designed by John J. Faillace, MD

Delicate enough to use on metacarpals but
strong enough for distal radius and larger
bones with its extra long ratchet
PRODUCT NO:

1171

Overall Length: 5" (12,7 cm)
Jaw Length: 1" (2,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTS

15

Small Bone Holding Forceps
with Long Ratchet
Designed for use in stabilizationof a fracture or osteotomy
PRODUCT NO:

1170

Overall Length: 5.75" (14,6 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANYO’Brien Bone Clamp
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
PRODUCT NO:

1816

Overall Length: 5.25" (13,3 cm)

OrthoLucent™
O’Brien Bone Clamp
Designed by Todd O’Brien, DPM

Designed for use in stabilizationof a fracture or osteotomy
The carbon fiber PEEK material is strong, lightweight,
completely radiolucent, can be steam sterilized, andhelps to prevent from marring component surfaces.
PRODUCT NO:

1815-R

Overall Length: 5.25" (13,3 cm)

MADE EXCLUSIVELYFOR INNOMED IN

SWITZERLAND

Lewin Small Bone Clamp
PRODUCT NO:

4685

Overall Length: 5" (12,7 cm)

16

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Pointed Fracture Reduction Clamps

Straight Left & Right

Designed by Reza Firoozabadi, MD MA

Curved Left, Straight Right

Versatile set of fracture reduction clamps, each witha specific tine design that allows for appropriate
vector placement so that anatomic reduction can beobtained in a number of different types of fractures
1.9 mm tines allow for a snug fit in 2 mm drill holes
Tines angled to prevent clamp “slippage” with compression
Straight tines can be placed deep within bone which allows for far
cortex compression.
Clamps incorporate a box joint design that prevents clamp joint
loosening and the need for tightening.
Example applications: any transverse fracture (straight-straightclamp), both bone forearm fractures, olecranon fractures, medial
malleolus fractures, and many more.
Speed Lock Style: Extra-long spin down allows for increased rangeof clamp use, and open-topped joint rotates to allow for increased
range of opening, and also allows for quick release
PRODUCT NO’S:
SMALL WITH SPEED LOCK MECHANISM

3666 [Straight Left & Right]

Small with
Speed Lock

Overall Length: 5.5" (14 cm)

3667 [Curved Left & Right]

Medium with
Speed Lock

Small with
RatchetOverall Length: 5.5" (14 cm)

3666-L [Curved Left, Straight Right]
Overall Length: 5.5" (14 cm)

3666-R [Straight Left, Curved Right]
Overall Length: 5.5" (14 cm)

SMALL WITH RATCHET MECHANISM

Straight Left & Right #3666
Curved Left & Right #3667
Curved Left, Straight Right #3666-L
Straight Left, Curved Right #3666-R

3668 [Straight Left & Right]
Overall Length: 5.5" (14 cm)

3669 [Curved Left & Right]
Overall Length: 5.5" (14 cm)

3668-L [Curved Left, Straight Right]

Straight Left & Right #3666-01
Curved Left & Right #3667-01
Curved Left, Straight Right #3666-L-01
Straight Left, Curved Right #3666-R-01

Straight Left & Right #3668
Curved Left & Right #3669
Curved Left, Straight Right #3668-L
Straight Left, Curved Right #3668-R

Overall Length: 5.5" (14 cm)

3668-R [Straight Left, Curved Right]

Two styles — Speed Lock and Ratchet — each available in four tine configurations

Overall Length: 5.5" (14 cm)

MEDIUM WITH SPEED LOCK MECHANISM

3666-01 [Straight Left & Right]
Overall Length: 7" (17,8 cm)

3667-01 [Curved Left & Right]
Overall Length: 7" (17,8 cm)

3666-L-01 [Curved Left, Straight Right]
Overall Length: 7" (17,8 cm)

3666-R-01 [Straight Left, Curved Right]

Straight
Left & Right

Overall Length: 7" (17,8 cm)

CurvedLeft & Right

Curved Left,
Straight Right

Straight Left,
Curved RightMedial Malleolar/Bone Fragment Clamps
Designed by Edward L. Sclamberg, MD

Quick tightening & release low profile
clamp with unlimited settings
PRODUCT NO’S:

1830 [Standard]

Overall Length: 5.5" (14 cm)
Clamp End Length: 1"

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY1835 [Medium]

Overall Length: 6" (15,2 cm)
Clamp End Length: 2"

1840 [Large]

Overall Length: 8" (20,3 cm)
Clamp End Length: 3"

Large

#1840

1.800.548.2362       FEBRUARY 2024   

Medium#1835

Standard#1830    FOOT & ANKLE INSTRUMENTS

17

Desai Jones Fracture Reduction Clamp
Designed by Sarang Desai, DO

Designed to reduce and maintain reductionof Jones fractures, helping to prevent
distraction and/or rotation during wire,
tap, and subsequent screw placement
Distally there are two K-wireholes for placement in the
distal 5th metatarsal and the
2-pronged clamp proximally isplaced on the tuberosity, allowing
a "high and inside" screw
placement without interference.

PRODUCT NO:

1802

Overall Length: 6" (15,2 cm)
Wire Block Length: 20 mm
Hole Separation: 5 mm on Center

Stanton Articulating
Small Bone Clamps
Designed by John L. Stanton, MD

Compress with a towel clamp

Distract with a gelpi retractor

Opposing clamps
facilitate manipulationof fracture ends
The small tube allows use of
a towel clamp to compress
non-union and shortening
osteotomies during fixation, as
well as to allow the use of Gelpi
retractors to distract malunions
during revision surgery.

PRODUCT NO’S:

1811-00 [Set of Two]
Also available individually:

1811-L [Left]

Right #1811-R

Overall Length: 5.125" (13 cm)
Curved Plate Radius: 5 mm
Pin Hole for Pins Up To: 2,4 mm

1811-R [Right]

Overall Length: 5.125" (13 cm)
Curved Plate Radius: 5 mm
Pin Hole for Pins Up To: 2,4 mm

Left #1811-L

Ludloff/Mau Osteotomy Fixation Clamp

Used after lateral hallux valgus correction of themetatarsal, the clamp allows for osteotomy fixationand cannulated screw guide wire direction
Clamp fixates the osteotomy to hold the correction, and the 15° slanted cannulated
K-wire guide allows the surgeon to place the guide wire for the cannulated screw
perpendicular to the osteotomy for final fixation of the osteotomy.
PRODUCT NO:

1812

Cannula Accepts K-wire up to: .045" (1,1 mm)
Overall Length: 5" (12,7 cm)

MADE EXCLUSIVELYFOR INNOMED IN

Designed by A. Austin

GERMANY

Calvo Medial Malleolus
Fracture Clamp
Designed by Ignacio Calvo, MD

Designed to reduce and hold adisplaced medial malleolus fracture

Right #1801-R

Left #1801-L

Also very useful in olecranon fractures.
PRODUCT NO’S:

1801-L [Left]
1801-R [Right]

18

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Chen Low Profile Plate/Bone Clamp
Designed by Franklin Chen, MD

The rotating nut is used totighten/release the center arm
once in the desired position

Designed for plate tobone clamping in avariety of lower and upper
extremity fractures
Useful for diaphyseal forearm fractures,
humerus fractures, and distal radius fractures.

The freely swivelingcenter arm allows foreasy placement, as well
as for quick release, aftergetting the legs in position

PRODUCT NO:

1639

Overall Length: 2.75" (7 cm)
Prong Depth: .675" (17 mm)
Prong Width: 5 mm

SMALLER SIZE

Redler Percutaneous Pin Clamp

Holds a small bone in apposition duringpercutaneous pinning of a fracture
Designed with a proximal pin tube with teeth; the tube guides
the pin and the teeth help keep the tube in place on the bone.
The distal tip is used to control the bone fragment. Includes along ratchet for locking on various sized bones, from 1 mm to 14
mm. Also useful during insertion of cannulated screw guide wires.
PRODUCT NO’S:

Three Tube Sizes
Available

Designed by M.R. Redler, MD

Overall Length: 5" (12,7 cm)
1810-35 Tube Diameter: .035" (.9 mm)
1810-45 Tube Diameter: .045" (1.1 mm)
1810-62 Tube Diameter: .062" (1.6 mm).035" (.9 mm) Tube Diameter #1810-35.045" (1.1 mm) Tube Diameter #1810-45.062" (1.6 mm) Tube Diameter #1810-62

Chang Pin Clamp
Designed by Win Chang, MD

Designed to allow accurate insertionof pins for internal fixation
Used for small bones, the clamp allows
accurate insertion of pins for internal fixation.
The cannula has a 1.8 mm internal diameter.
PRODUCT NO:

1760-01

Cannula Internal Diameter: 1.8 mm
Overall Length: 6" (15,2 cm)
Locking Ratchet Opens To: 25 mm(Prototype used in
X-ray images)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Teurlings Medial Malleolar
Clamp with Wire Guide

Designed by Luc Teurlings, MD

Helps to stabilize the medial malleolar
fragment during internal fixation
MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTSPRODUCT NO:

1803

Cannula Diameter: .062" (1.6 mm)
Overall Length: 5.25" (13,3 cm)

19

Mogul K-Wire/Pin Insertion Guide
Designed by Stuart J. Mogul, DPM, FACFAS

A guide designed for passing guidepins or K-wires through two adjacentmetatarsal bones
PRODUCT NO:

3017

Dimensions: 2.375" Tall x 3.75" Wide (6 x 9,5 cm)
Maximum Pin Diameter: 3/32" (2,4 mm)
Maximum Clamped Opening: 2" (5,1 cm)
Minimum Clamped Opening: .375" (1 cm)
Pin/K-Wire Guide Length: .925" (23,5 mm)

Zell Fixed Angle Wire Guide
Designed by Richard Zell, MD

Designed to help with placement ofguide wires for cannulated screws
and k-wires in foot and ankle
surgeries, such as bunion surgery,midfoot fusion, and midfoot ORIFPRODUCT NO:

3021

Overall Length: 2.75" (7 cm)
Handle Platform: 1" x .875" (25 x 22 mm)
Guide Tube Angle: 35°
Guide Tube for wires up to: .052"/1.3 mm

Extended Drill Sleeves

Designed by Reza Firoozabadi, MD

Designed to help reduce fractures when K-wires are passed through,the extra long drill sleeve helps to protect soft tissues and preventthe need for stacking two drill sleeves
2.4/1.8 mm #3014-01

Serrated tips allow for better grip
when drilling at an angle or whenpushing a fracture fragment toassist with fracture reduction
Sleeve can be used as a reduction
aid with placement of a kirschner
wire through sleeve
Collaborated tips which allow
placement of appropriate size drills
for lagging by technique — as an
example a 2.5 end will fit into a
3.5 drill hole

2.7/2.0 mm #3014-02
PRODUCT NO’S:

3014-00 [Set of Three]
Set Includes / Available Individually:

3.5/2.5 mm #3014-03

3014-01 [2.4/1.8 mm]

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

Case example of using modified
3.5/2.5 mm drill sleeve for
placing 3.5 mm screws in a
forearm fracture case. Note
how extended sleeve protects
soft tissues during drilling.

3014-02 [2.7/2.0 mm]

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

Profile3014-03 [3.5/2.5 mm]

Extra Long Guide Sleeve Length: 2.25" (5,7 cm)

Larger Diameter

Serrated Sleeve Ends

20

Collaborated tip end diameters
allow placement of appropriate size drills
for lagging by technique — as an example a
2.5 end will fit into a 3.5 drill hole.

Overall Length: 6.875" (17,6 cm)
Guide Tube Length: 2.25" (5,7 cm)
Guide Angle from Handle: 30°

Smaller Diameter

Serrated Sleeve Ends

WWW.INNOMED.NET   

Note 2.0 mm end of drill
sleeve placed into a predrilled
2.7 mm hole, utilized as a lag
by technique 2.7 mm screw.

   FREE TRIAL ON MOST INSTRUMENTS

Resnick Allis Bone Clamp
Designed by Charles T. Resnick MD

A traditional Allis Bone Clamp
designed with a longer ratchet
which allows for a wider opening
to allow a bone to be clamped
and locked onto
PRODUCT NO:

1385

Overall Length: 6" (15,2 cm)
Ratcheted Clamp Opens to: 37 mm
Clamp End Width: 4.7 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Coated Allis Bone Clamps

A traditional Allis Bone Clamp designed witha longer ratchet—for a wider opening to allow
a bone and plate to be clamped and locked
onto—and coated end(s) to prevent frommarring a component surface
PRODUCT NO’S:

1381 [One Coated End]One Coated End #1381

Modification of design by Charles
T. Resnick MD

Overall Length: 6.125" (15,9 cm)
Ratcheted Clamp Opens to: 35 mm
Non-coated-end Width: 4 mm

Two Coated Ends #1382

1382 [Two Coated Ends]

Overall Length: 6.125" (15,9 cm)
Ratcheted Clamp Opens to: 35 mm
Non-coated-end Width: 4 mm

Slavitt Phalangeal Forceps

Designed by Jerome Slavitt, DPM

Designed to enable the surgeon to provide joint
distraction and stability during joint placement at thebase of the proximal phalanx of the lesser digits
Helps to distract the joint and hold the bone,allowing easier access to the base. Can also
be used for digital fusions to hold bones
better for drilling and cutting applications.
PRODUCT NO:

1163

Overall Length: 6" (15,2 cm)
Clamp Internal Opening Diameter: 4 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Faillace Bone Impact/Graft Forceps
Design modification by John J. Faillace, MD, FAAOSLong vertical grooves at the tip are designed to deliver graft
into a small space, where a freer elevator can be used topush the graft down into the space, then the closed
flat end can be used to tamp down the graft
PRODUCT NO:

5011

Overall Length: 5" (12,7 cm)
Tip Diameter When Closed: 3,2 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTS

21

Universal Bone Grafting/Impacting Forceps

Designed by J. A. Amis, MD

Bone graft can be grasped, placed & impacted
without changing hands or instruments

1/8"
3/16"
1/4"
5/16"
(3,2 mm) (4,8 mm) (6,3 mm)(8 mm)
Diameter ends at actual size (closed forceps)

The forceps are designed with grasping ends for delivery
of bone graft. When the graft is in place, the forceps are closed,
which forms the ends into an impacting punch. A striking platform
is attached to the end of the forceps for tapping and tamping the graft.
Four end diameters are available in two lengths.
PRODUCT NO’S:
Short: 6" (15,2 cm) Length

5010-01 1/8" (3,2 mm) Diameter End
5010-02 3/16" (4,8 mm) Diameter End
5010-03 1/4" (6,3 mm) Diameter End
5010-04 5/16" (8 mm) Diameter End

Long: 10" (25,4 cm) Length

5050-01 1/8" (3,2 mm) Diameter End
5050-02 3/16" (4,8 mm) Diameter End
5050-03 1/4" (6,3 mm) Diameter End
5050-04 5/16" (8 mm) Diameter End

When the forceps areclosed, they form into
an impacting punch

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

Sarraf TiN Coated
Cement Removal Forceps

Straight #5039

Designed by Khaled M. Sarraf, MD

Ultra hard titanium nitride coating helps to extend
forceps life by increasing surface hardness,prolonging sharpness, and resisting chemicals
and corrosion, while helping to eliminate metaltransfer and protect the implant surface.
PRODUCT NO’S:

5039 [Straight]

Overall Length: 6" (15,2 cm)

Bent #5041

5041 [Angled]

Overall Length: 6.125” (15,6 cm)

The curved semicircular tip is congruent
to most tibial plates and femoral condylar
implants, helping to facilitate removalof excess cement, especially at the tightposterior aspect
The small scoop-end tip assists in excising
unset cement
Ultra hard titanium nitride coating helps
to extend curette life by increasing surface
hardness, prolonging sharpness, and resistingchemicals and corrosion, while helping toeliminate metal transfer and protect the
implant surface

Sarraf Cement Trimmer
Designed by Khaled M. Sarraf, MD

Two-in-one instrument designed for cement
removal during arthroplasty surgery

Bozeman Cement Trimmer

PRODUCT NO:

5212

Overall Length: 7.75" (19,7 cm)

Combines the two most common cement trimming tools into one

Designed by Daniel M. Gannon, MD

The tool has a blunt blade tip on one end to help with separation of
the trimmed cement. The angled curette end helps gather the trimmed
cement. The thin shank and angled curette can reach into tight spaces
such as the back of the implants to remove excess cement. The ends
are titanium nitrite coated to help eliminate metal transfer.

PRODUCT NO:

5245

Overall Length: 8.5" (21,6 cm)

22

WWW.INNOMED.NET   

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY   FREE TRIAL ON MOST INSTRUMENTS

K-Wire Bender/Cutter
1 and
1.2 mm

1.2 and
1.6 mmThe right slot of the
instrument’s lower
jaw can hold K-Wireswith a diameter of
1.2 mm or 1.6 mm.
The smaller left slot
can hold K-Wires
measuring 1 mm or
1.2 mm in diameter.

Bending

Smooth Bending
Clean Cutting

Designed to bend a K-Wire while extending
from bone without applying mechanical strain
The K-Wire only needs to extend 20 mmfrom the skin surface to be bent.

Can bend and cut K-Wires
measuring 1 to 1.6 mm(.039–.062") in diameter
PRODUCT NO:

2111

Overall Length: 6.5" (16,5 cm)

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

With the jaw of the instrument
opened wide, the K-Wire isinserted from the side into oneof the slots of thelower jaw. During bending, theK-Wire is forced backwards by thenose of the upper jaw and guided
by a small groove.

Cutting

The K-Wire is inserted into
the cutting groove and thebender/cutter cuts by shearing(like a cigar cutter), notcrushing. The result is a clean
and burr-free cut surface.

Stanton Straight Pin Removal Pliers
Designed by John Stanton, MD
PRODUCT NO:

1893

Overall Length: 6.375" (16,2 cm)
Jaw Length: 1.62 (4,1 cm)
Instrument Width: 1 cm

Pin Puller - Small

Small size allows for use in asmall incision to help with removalof a 2 mm or smaller K-wire pin

PRODUCT NO:

3033

Overall Length: 6.5" (16,5 cm)
Jaw Width: 6,2 mm tapering to 3 mm at end
Jaw Height: 11,7 mm

Stanton Bent Pin Removal Pliers
Designed by John Stanton, MD, FACS
PRODUCT NO:

1894

Overall Length: 6.5" (16,5 cm)
Jaw Length: 1.65" (4,2 cm)
Instrument Width: 1 cm

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTS

23

DesaiCurette Osteotomes

Cup Sizes:
5 x 6 mm

Designed by Sarang Desai, DO

The osteotome portion also
can be used to "feather"
the subchondral surface toexpose bleeding bone. It isalso useful in instances ofobtaining autograft, as it canbe used to create a bonewindow and then remove
cancellous bone.

8 x 10 mm

5 x 6 mm
#5241

PRODUCT NO’S:

8 x 10 mm

5241 [5 x 6 mm]

Overall Length: 8.25" (21 cm)
Osteotome Width: 3.5 mm
Osteotome Length: 3.5 mm from edge of cup

5242 [8 x 10 mm]

Overall Length: 8.25" (21 cm)
Osteotome Width: 6.5 mm
Osteotome Length: 3 mm from edge of cup

Designed to remove bone andcartilage, helpful for preparing joint surfaces
for fusion, allowing easy removal of osteophytes
and cartilage without having to switch instruments#5242

Corkscrew Small Bone Manipulator
Designed by Raymond Wurapa, MD

Designed with an aggressive thread to aid in
excising small bones of the hand and footThe quick-connect end allows the device to be inserted with ease
under power with a standard drill attachment. After insertion, thedrill is detached and manual control over the process of extracting
the bone can be performed by hand, using either the disc on theshaft or attaching a handle.
Helps with removal of trapezium during basal joint arthroplasty
Helps with extraction of any carpal bones forwrist procedures: proximal row carpectomy (PRC),
partial wrist fusions, pisiform excision

PRODUCT NO:

1615

Overall Length: 4" (10,2 cm)
Length Beyond Disc: 2.25" (5,7 cm)
Length Beyond Line: .625" (1,6 cm)
Corkscrew Length: .375" (1 cm)
Optional:

Shown with optional
universal handle
attached(Handle not included.)

Lubahn Corkscrew
Designed to help
with removalof tarsal
and/or carpal
bonesStandard#1191

Aids trapezium removal during basal joint
arthroplasty when the bone is being removed as a unit
Can also be used to facilitate a proximal row carpectomy
as it fits the scaphoid, lunate, and triquetrum
May additionally be used to remove the pisiform in casesof arthritis of the pisotriquetral joint

24

Overall Length: 4" (10,2 cm)

Prototype Shown

Designed by John D. Lubahn, MD

Prototype
Shown

S0113 [Universal Handle]

PRODUCT NO'S:

1191 [Standard]

Overall Length: 2.25" (5,7 cm)

Extended#1191-01

WWW.INNOMED.NET   

1191-01 [Extended]

Overall Length: 6.5" (16,5 cm)   FREE TRIAL ON MOST INSTRUMENTS

Ring Curettes – Straight Shaft

Ring Curettes – Bent Shaft

3 mm Straight #5150

3 mm Bent #5156

6 mm Straight #5152

6 mm Bent #5157

8 mm Straight #5154

8 mm Bent #5158

PRODUCT NO’S:
Straight Shaft

Overall Length: 8.75" (22,2 cm)

5150 [3 mm Straight]
Ring Diameter: 3 mm

PRODUCT NO’S:
Bent Shaft

MADE FOR INNOMED IN

GERMANY

5152 [6 mm Straight]
Ring Diameter: 6 mm

5156 [3 mm Bent]

5154 [8 mm Straight]

Ortho Mini Gouges

MADE FOR INNOMED IN

Overall Length: 8.625" (21,9 cm)
Ring Diameter: 3 mm

Ring Diameter: 8 mm

2 mm Gouge #1168-2
3 mm Gouge #1168-3
4 mm Gouge #1168-4
5 mm Gouge #1168-5
6 mm Gouge #1168-6
7 mm Gouge #1168-7
8 mm Gouge #1168-8

GERMANY

5157 [6 mm Bent]

Ring Diameter: 6 mm

5158 [8 mm Bent]

Ring Diameter: 8 mm

PRODUCT NO’S:

1168-2 [2 mm Gouge]

Overall Length: 5.75" (14,6 cm)
Gouge Width: 2 mm

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY

1168-3 [3 mm Gouge]

1168-6 [6 mm Gouge]

Overall Length: 5.75" (14,6 cm)
Gouge Width: 3 mm

Overall Length: 5.75” (14,6 cm)
Gouge Width: 6 mm

1168-4 [4 mm Gouge]

1168-7 [7 mm Gouge]

Mini orthopedic gouges
with ergonomic handles, designed forbone resection in small areas and resection of periosteum

1168-5 [5 mm Gouge]

1168-8 [8 mm Gouge]

Ortho Mini Chisels

PRODUCT NO’S:
Offset Chisels

Mini orthopedic chisels, straight and offset,with straight and ergonomic handles

Overall Length: 5.75" (14,6 cm)
Gouge Width: 4 mm
Overall Length: 5.75” (14,6 cm)
Gouge Width: 5 mm

1 mm Offset Chisel #1169-1
2 mm Offset Chisel #1169-2
3 mm Offset Chisel #1169-3
4 mm Offset Chisel #1169-4
5 mm Offset Chisel #1169-5

Overall Length: 5.75" (14,6 cm)
Gouge Width: 7 mm
Overall Length: 5.75" (14,6 cm)
Gouge Width: 8 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 1 mm

Overall Length: 6.4" (16,3 cm)
Chisel Width: 3 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 2 mm

Overall Length: 6.4" (16,3 cm)
Gouge Width: 4 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 3 mm

Overall Length: 6.4" (16,3 cm)
Gouge Width: 5 mm

Overall Length: 6.25" (15,9 cm)
Chisel Width: 4 mm

MADE EXCLUSIVELYFOR INNOMED IN

1169-2 [2 mm Offset Chisel] 1170-4 [4 mm Straight Chisel]1169-3 [3 mm Offset Chisel] 1170-5 [5 mm Straight Chisel]1169-4 [4 mm Offset Chisel]

3 mm Straight Chisel #1170-3
4 mm Straight Chisel #1170-4
5 mm Straight Chisel #1170-5

Micro Curettes
Four cup sizes,
straight or 45°
angled-end shaft

1169-5 [5 mm Offset Chisel]

GERMANY

Overall Length: 6.25" (15,9 cm)
Chisel Width: 5 mm

Straight, Cup Size 6/0 #4246
Straight, Cup Size 4/0 #4244
Straight, Cup Size 1 #4240
Straight, Cup Size 2 #4242
Angled, Cup Size 6/0 #4246-01
Angled, Cup Size 4/0 #4244-01
Angled, Cup Size 1 #4240-01
Angled, Cup Size 2 #4242-01

1.800.548.2362       FEBRUARY 2024   

Straight Chisels

1169-1 [1 mm Offset Chisel] 1170-3 [3 mm Straight Chisel]    FOOT & ANKLE INSTRUMENTSPRODUCT NO’S:
Straight Micro Curettes

Overall Length: 9.75" (24,8 cm)
Shaft Length: 4.5" (11,4 cm)
4242 Cup Size 24240 Cup Size 14244 Cup Size 4/04246 Cup Size 6/0
Angled Micro Curettes

Overall Length: 9.75" (24,8 cm)
Shaft Length: 4.5" (11,4 cm)
4242-01 Cup Size 24240-01 Cup Size 14244-01 Cup Size 4/0
4246-01 Cup Size 6/025Flexible Osteotome Instruments
An assortment of flexible osteotome blades
useful in foot & ankle surgery procedures

Choice ofHandle Style
Handle with Quick-Coupling End #S1020

Handle with Locking Nut #S1021

2.5" (6,4 cm) x 8 mm Osteotome Blade #S1002

Sharp, flexible blades are well suited for loosening implants from
cement or bony ingrowth fixation
Various blade widths and profiles allow great flexibility to follow the
implant contours
Modular handle is made of high impact surgical stainless steel andhas a quick-coupling positive locking mechanism for ease of use andquick blade changes
Slap hammer threads into the handle and is designed to facilitate
blade removal
Optional Strike Plate can be attached to the Handle for direct striking
with a mallet
Optional Curved Chisel Blades can be used to help loosen the cement/
prosthesis interval in total ankle revisions. The curved design is useful
in working around pegs & fins to get posterior cement access. Also
helpful with removal of other implants, i.e shoulder, knee, femoral, etc.

2.5" (6,4 cm) x10 mm Osteotome Blade #S1003
2.5" (6,4 cm) x12 mm Osteotome Blade #S1004
2.5" (6,4 cm) x 20 mm Osteotome Blade #S1005
2.5" (6,4 cm) x 12 mm Curved Osteotome Blade #S1006

2.5" (6,4 cm) x 8 mm Chisel Blade #S1222
2.5" (6,4 cm) x 10 mm Chisel Blade #S1223
2.5" (6,4 cm) x 12 mm Chisel Blade #S1224
2.5" (6,4 cm) x 20 mm Chisel Blade #S1225

PRODUCT NO’S:
Individual Instruments Available Separately

S1002 [Osteotome Blade] 2.5" (6,4 cm) x 8 mm
S1003 [Osteotome Blade] 2.5" (6,4 cm) x 10 mm
S1004 [Osteotome Blade] 2.5" (6,4 cm) x 12 mm
S1005 [Osteotome Blade] 2.5" (6,4 cm) x 20 mm
S1006 [Curved Osteotome Blade] 2.5" (6,4 cm) x 12 mm
S1020 [Handle with Quick-Coupling End] 5" (12,7 cm)
orS1021 [Handle with Locking Nut] 5" (12,7 cm)
S1020-SP [Strike Plate for Handle] Diameter 1.625" (4,1 cm)
S1222 [Chisel Blade] 2.5" (6,4 cm) x 8 mm
S1223 [Chisel Blade] 2.5" (6,4 cm) x 10 mm
S1224 [Chisel Blade] 2.5" (6,4 cm) x 12 mm
S1225 [Chisel Blade] 2.5" (6,4 cm) x 20 mm
S1228 [Chisel Blade] 5" (12,7 cm) x 10 mm
S1233-L [Left Curved Chisel Blade] 2" (5,1 cm) x 8 mm
S1233-R [Right Curved Chisel Blade] 2" (5,1 cm) x 8 mm
S2007 [Slap Hammer] 12" (30,5 cm)

Curved chisel design
allows working
around component
pegs, fins, etc.

2" (5,1 cm) x 8 mm Left Curved Chisel Blade #S1233-L
2" (5,1 cm) x 8 mm Right Curved Chisel Blade #S1233-R

Bevel side awayfrom component

Slap Hammer #S2007

Complete Set with more
options available onlineat www.innomed.net

Strike Plate for Handle #S1020-SP

Medial and Lateral Curve Radial Blades designed by Henry Boucher, MD
Curved Chisel Blades designed by William McMaster, MD

Narrow Cement Removal Gouge, Short #S7505

Mueller-Type Cement
Removal Instruments

Useful for cement removal in the ankle
Also helpful in hip, knee, and shoulder surgery.
PRODUCT NO’S:
Individual Instruments Available Separately

Offset Chisel #S7520

S7505 [Narrow Cement Removal Gouge, Short]
Shaft Length: 10 cm
Gouge: 9 mm, negative

S7520 [Offset Chisel]

Shaft Length: 15 cm
Chisel: 9 mm

Cement Removal Osteotome, Short #S7595

Shaft Length: 15 cm
Osteotome: 8 mm

S7540 [4.4 mm Drill]
S7545 [4.4 mm Drill Guide]
S7570 [Cross Bar]

4.4 mm Drill #S7540

Complete Set
with more options
available online atwww.innomed.net

4.4 mm Drill Guide #S7545
Cross Bar #S7570

26

S7595 [Cement Removal Osteotome, Short]

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Prototype
Shown

Nicholson Small Bone andShoulder Cement Removal Instruments

Designed by Gregory Nicholson, MD

Designed to facilitate cement removal in smaller diameter boneof the humerus, ulna, and smaller implant geometries
Reverse bevel tip helps the gouge to slide between thebone and cement
T-shaped Gouge-Splitter allows the gouge to slide between
the cement and bone and vertically split the cement
mantle to facilitate removal
Small diameter widths and curvatures more closely matchshoulder and elbow implants and smaller bone diameters
Shorter length allows for better control and access

Extra Small 5 mm Gouge #5251-05
Small 7 mm Gouge #5251-07
Medium 9 mm Gouge #5251-09

PRODUCT NO’S:

Gouges Overall Length: 9" (22,9 cm)
Gouges Handle Length: 4" (10,2 cm)

Large 11 mm Gouge #5251-11

5251-00 [Complete Set w/Case
Set Includes / Available Individually:

5251-05 [Extra Small]

Small 7 mm Gouge with Splitter #5252-07

Gouge Width: 5 mm

5251-07 [Small]
Gouge Width: 7 mm

5251-09 [Medium]

Medium 9 mm Gouge with Splitter #5252-09

Gouge Width: 9 mm

5251-11 [Large]
Gouge Width: 11 mm

Large 11 mm Gouge with Splitter #5252-11

5252-07 [Small w/Splitter]
Gouge Width: 7 mm
Splitter Height: 4 mm

5252-09 [Medium w/Splitter]
Gouge Width: 9 mm
Splitter Height: 5 mm

Backhook #5254

5252-11 [Large w/Splitter]
Gouge Width: 11 mm
Splitter Height: 6 mm

5254 [Backhook]

Overall Length: 12.5" (31,8 cm)
Handle Length: 4.5" (11,4 cm)
Shaft Diameter: 4 mm

Footed Impactor #5255

5255 [Footed Impactor]

Foot Pad Size: 8.5 mm x 11.5 mm
Shaft Diameter: 8.5 mm (21,6 cm)
Overall Length: 12.75" (32,4 cm)
Handle Length: 4.5" (11,4 cm)

Used to help remove a humeral implant by impacting the medial collar of theprosthesis — helps provide a very direct parallel force to the implant for removal

5253 [Case for Set

AndersonTalar Neck Osteotome

9.5 mm#5075-75

12.7 mm#5075-50

17 mm

#5075

Designed by John Anderson, MD

Designed to help improve rangeof motion and reduce pain caused byanterior boney impingement of the ankle
by removing osteophytes from the anterior talar
neck and the anterior distal tibia
PRODUCT NO’S:

5075

Osteotome Width: 17 mm
Overall Length: 9.875" (25,1 cm)
Handle Length: 4.5" (11,4 cm)

5075-50

Osteotome Width: 12.7 mm
Overall Length: 9.875" (25,1 cm)
Handle Length: 4.5" (11,4 cm)

5075-75

Osteotome Width: 9.5 mm
Overall Length: 9.875" (25,1 cm)
Handle Length: 4.5" (11,4 cm)

1.800.548.2362       FEBRUARY 2024       FOOT & ANKLE INSTRUMENTS

27

Mini-lexer Gouges

4 mm Gouge #2022-02

PRODUCT NO’S:

2022-02 [Mini Lexer Gouge – 4 mm]
Overall Length: 7" (17,8 cm)
Gouge Width: 4 mm

2022-03 [Mini Lexer Gouge – 6 mm]

6 mm Gouge #2022-03

Overall Length: 7" (17,8 cm)
Gouge Width: 6 mm

2022-04 [Mini Lexer Gouge – 10 mm]
Overall Length: 7" (17,8 cm)
Gouge Width: 10 mm

10 mm Gouge #2022-04

MADE FOR INNOMED IN

GERMANY

Mini-lexer Osteotomes

Small, thin osteotomes helpful in osteophyte
and cement removal in total joint surgery.
Larger handle helps with better control.

Helpful in osteophyte and cement removal
PRODUCT NO’S:

5270-01

5270-035270-025270-04

Blade Width: 4 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

12 mm#5270-04

10 mm#5270-03

Blade Width: 6 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

6 mm#5270-02

Blade Width: 10 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)
Blade Width: 12 mm
Overall Length: 7.25" (18,4 cm)
Handle Length: 4" (10,2 cm)

MADE FOR INNOMED IN

GERMANY

4 mm#5270-01

Roberts Pin Bending Cannula Set
Designed by David Roberts, MD

Helpful when removing a pin
which has been cut flush to thebone which can be hard to grasp
with standard tools.
PRODUCT NO’S:

2113-00 [Set of 3 Sizes]
Set Includes / Available Individually:

2113-01 [2 mm]

For 1.5 - 2.0 mm flexible nails
Overall Length: 5.5" (14 cm)

Designed to help bend the end of a flexible
intramedullary pin, which has been cut flush to thebone, for better grasping during pin removal
After exposing the pin end, the cannula helps bend
the pin for better access for the removal
instrument while maintaininga small incision.
4 mm #2113-03

2113-02 [3 mm]

For 2.5 - 3.0 mm flexible nails
Overall Length: 5.5" (14 cm)

Features
Beveled edge that allows
easy capture of end of pin
Cannula can be used as a
trephine for pins with bonyovergrowth
Reusable - thick walls
withstand repeated uses

3 mm #2113-02
2 mm #2113-01

2113-03 [4 mm]

For 3.5 - 4.0 mm flexible nails
Overall Length: 5.5" (14 cm)

Technique

Distal Radius – 1.5 cm Incision

Proximal Ulna – 1.5 cm Incision

Advance cannula over the exposed pin.
The beveled end makes this easier andcan be used as a trephine in when bonyovergrowth is present.
Slowly use the tool to bend theend of the pin away from bone, whichhelps with pin removal.
The bent end can be removed
with standard pliers.
28

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Chandran
Double Ball Spike
Designed by Rama E. Chandran, MD

Designed to help rotate and control
a butterfly bone fragment for fixation
PRODUCT NO:

8027

Overall Length: 12.5" (31.8 cm)
Handle Length: 4.625" (1,17 cm)

Resnick Small Bone Tamp with Oblique K-Wire Hole
Designed by Charles Resnick, MD

Design allows for the concurrent reduction of a fracture
and placement of a wire into the fracture site —especially helpful when the surgical exposure issmall and tight, the fracture fragments are
small, and the reduction is demanding
TWO SIZES AVAILABLE:
Wire Hole for K-wires up
to 1.1 mm (.045”) or
1.6 mm (.062”)

1.2 mm Hole #5294
1.6 mm Hole #5294-01
BeforeAfter

PRODUCT NO’S:

5294 [1.2 mm Hole]

Wire Hole for: 1,2 mm (.045") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

The serrated distal end minimizes slippage on the cortical surface, does notinterfere with the placement of the guidewire and allows for subsequent surgeondecided, intraoperative angulation of the wiring once the first cortex is drilled
Especially useful in fractures where there is involvement of an articular surface,for example, mallet fractures of the distal phalanx, articular fractures that involve
ligamentous attachments or tendon attachments of the phalanges, scaphoid polesmall fracture fragments or other small carpal fractures, and radial styloid fractures

5294-01 [1.6 mm Hole]

Wire Hole for: 1,6 mm (.062") K-wire
Overall Length: 7.5" (19,1 cm)
Shaft Diameter: 6,3 mm
End Diameter: 2,5 mm

Rudisill Locking
Small Bone Reduction Forcep
Designed by Ed Rudisill, MD

PRODUCT NO:

For reduction of hand phalanx
and metacarpal fractures

Small Cannulated Ball Spike

MADE EXCLUSIVELYFOR INNOMED IN

GERMANY2017

Overall Length: 4.875" (12,4 cm)

Designed by Benjamin C. Taylor, MD

Designed to help reduce a bone fragment
and keep it reduced, while the cannulation
allows placement of a K-wire (up to 1.6
mm/.062") into the fragment
Helps to prevent slipping while inserting K-wires
Can serve as a handle for K-wire joysticks

1.800.548.2362       FEBRUARY 2024   

PRODUCT NO:

8092

Overall Length: 4.5" (11,4 cm)
Handle Length: 3" (7,6 cm)
Ball Diameter: .275" (7 mm)    FOOT & ANKLE INSTRUMENTS

29

LawtonScrew Extractors
Designed by Jeffrey Lawton, MD

Designed to help
extract mini andmicro fragment
screws; small
cannulated screws;
or headless screws

PRODUCT NO’S:

7653-00 [Set w/Case]
Set Includes / Available Individually:

7653-01 [1.5 mm]

1.5 mmOverall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)#7653-01

2.5 mm#7653-02

3.5 mm#7653-037653-02 [2.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

7653-03 [3.5 mm]

Overall Length: 6" (15,2 cm)
Handle Width: 4" (10,2 cm)

1025 [Sterilization Case]

Lawton Broken Screw Extractor
Designed by Jeffrey Lawton, MD

Designed to help remove broken or
stripped screws (1 - 2 mm)
PRODUCT NO:

7653-04

Overall Length: 4" (10,2 cm)
Handle Width: 3" (7,6 cm)

Hooked Bone Awls

Standard hooked bone awl being used to gain length
to assist with reduction of a fibula fracture. A 2mm
pilot hole is made to seat the tip of the bone awl.

Designed by Reza Firoozabadi, MD“Shoulder hook” awls designed
to help with manipulation ofbone fragments for fixation

Standard #5078

PRODUCT NO’S:

5078 [Standard]

Overall Length: 10.5" (26,7 cm)
Handle Length: 5" (12,7 cm)

5078-01 [Long]

Overall Length: 13.375" (34 cm)
Handle Length: 6" (15,2 cm)

Long #5078-01

Fracture Reduction Pick

Used to align bone fragments,
and to pick away tissue andbone fragments
PRODUCT NO:

S0129

Overall Length: 6.25" (15,9 cm)

30

WWW.INNOMED.NET      FREE TRIAL ON MOST INSTRUMENTS

Overall Length: 8" (20,3 cm)

BasicScrew Removal System
System designed to help remove damaged and broken screws from 1.5 to 7.0 mm

Overall Length: 7" (17,8 cm)
Gouge Width: 4 mm

Screw Removal Pliers

PRODUCT NO’S:

2022-00 [Complete System with Case]
System Includes / Available Individually:

2022-01 [Screw Removal Pliers]
2022-02 [Mini Lexer Gouge – 4 mm]
2022-03 [Mini Lexer Gouge – 6 mm]
Overall Length: 7" (17,8 cm)
Gouge Width: 6 mm

2022-04 [Mini Lexer Gouge – 10 mm]
Overall Length: 7" (17,8 cm)
Gouge Width: 10 mm

2022-05 [Extraction Screwfor 1.5/2.0 mm Screw]
Overall Length: 1.6" (4 cm)

2022-06 [Extraction Screwfor 2.7/3.5/4.0 mm Screw]
Overall Length: 1.6" (4 cm)

2022-07 [Extraction Screwfor 4.5/5.0/6.5/7.0 mm Screw]
Overall Length: 1.6" (4 cm)

2022-CASE [Case for System]2022-IP [Instruction Plate]Dimensions: 7.875" x 3.65" (20 x 9,3 cm)

2022-SH [Sharp Hook]#2022-01

Sharp HookMini Lexer Gouges#2022-SH

4 mm Gouge #2022-02

T-Handle with A0-End
6 mm Gouge #2022-03#2022-T

Overall Length: 6.1" (15,5 cm)

2022-T [T-Handle with A0-End]Overall Length: 5.9" (15 cm)
Handle Width: 3.15" (8 cm)

2023-01 [Extraction Bolt
for 1.5 mm Screw]

For 1.5/2.0 mm #2022-05

Overall Length: 2.35" (6 cm)

For 4.5/5.0/6.5/7.0 mm #2022-07

Overall Length: 2.35" (6 cm)
Overall Length: 2.35" (6 cm)

Extraction Bolts
For 1.5 mm Screw #2023-01

2023-04 [Extraction Bolt
for 3.5/4.0 mm Screw]

For 2.0 mm Screw #2023-02

Overall Length: 2.35" (6 cm)

2023-05 [Extraction Bolt
for 4.5 mm Screw]

For 2.7 mm Screw #2023-03

Overall Length: 3.15" (8 cm)

For 3.5/4.0 mm Screw #2023-04

2023-06 [Extraction Bolt
for 5.0/6.5/7.0 mm Screw]
Overall Length: 3.94" (10 cm)

For 4.5 mm Screw #2023-05

2023-07 [Trephine for 1.5 mm Screw]
Overall Length: 4.125" (10,5 cm)

For 5.0/6.5/7.0 mm Screw #2023-06

2023-08 [Trephine for 2.0 mm Screw]
Overall Length: 4.125" (10,5 cm)

2023-09 [Trephine for 2.7 mm Screw]
Overall Length: 4.125" (10,5 cm)

Set in CaseFor 2.7/3.5/4.0 mm #2022-06

2023-02 [Extraction Bolt
for 2.0 mm Screw]2023-03 [Extraction Bolt
for 2.7 mm Screw]

10 mm Gouge #2022-04

Extraction Screws

Trephines

2023-10 [Trephine for 3.5/4.0 mm Screw]

For 1.5 mm Screw #2023-07

2023-11 [Trephine for 4.5 mm Screw]

For 2.0 mm Screw #2023-08

Overall Length: 4.125" (10,5 cm)
Overall Length: 5.4" (13,7 cm)

For 2.7 mm Screw #2023-09

2023-12 [Trephine for
5.0/6.5/7.0 mm Screw]

For 3.5/4.0 mm Screw #2023-10

Overall Length: 5.4" (13,7 cm)

2024-01 [Spare Trephine Cutting End
for 1.5 mm Screw]

For 4.5 mm Screw #2023-11

Overall Length: 1.6" (4 cm)

2024-02 [Spare Trephine Cutting End
for 2.0 mm Screw]
Overall Length: 1.6" (4 cm)

2024-03 [Spare Trephine Cutting End
for 2.7 mm Screw]
Overall Length: 1.6" (4 cm)

For 5.0/6.5/7.0 mm Screw #2023-12

Spare Trephine Cutting Ends
For 1.5 mm Screw #2024-01

2024-04 [Spare Trephine Cutting End
for 3.5/4.0 mm Screw]

For 2.0 mm Screw #2024-02

2024-05 [Spare Trephine Cutting End
for 4.5 mm Screw]

For 3.5/4.0 mm Screw #2024-04

Overall Length: 1.6" (4 cm)

Instruction Plate #2022-IP

Side AFor 2.7 mm Screw #2024-03

Overall Length: 2.75" (7 cm)

For 4.5 mm Screw #2024-05

2024-06 [Spare Trephine Cutting End
for 5.0/6.5/7.0 mm Screw]

For 5.0/6.5/7.0 mm Screw #2024-06

Overall Length: 2.75" (7 cm)

MADE FOR INNOMED IN

GERMANY

Side B

Measurements in this Catalog
All effort has been made to ensure the accuracy of themeasurements listed in this catalog, however, somesmall differences may exist between actual and listed
measurements.

Measurements of overall length are
the linear distance from one endof the product to the furthest
opposite end, as shownin these examples:

Measurements of
blade width are the linear
distance from one side of the productto the opposite side, typically at the widest
point, as shown in this example:

31

PRSRT. STD.
U.S. POSTAGE
PAID
BOURBON, IN

103 Estus Drive
Savannah, GA 31404

PERMIT NO. 29

FREE
TRIAL
on mos t ins trument s
Instruments are available for a no-charge two-week evaluation — includes FREE UPS Ground Shipping**When shipped to a hospital or medical center; additional charge applies for expedited shipping.
Free trial offer excludes implant extraction instruments, which are available as rentals. There is a pad replacement charge with the hip positioners.

Ratcheting Reduction Clamp Kit
Designed by Michael Craig, OPA-C

Designed as a soft
tissue sparing fracture
reduction clamp

PRODUCT NO’S:

Plate
Point

3840-00 [Clamp Kit]
Also available Individually:

Screw
Point

Percutaneous
Points

3840-02 [Plate Point]

Overall Length: 1" (2,54 cm)

3840-03 [Screw Point]

Overall Length: .875" (2,2 cm)

3840-04 [Percutaneous Point]

2 included in set, one with this product number
Overall Length: 1" (2,54 cm)

3840-MA [Ratcheting Reduction
Mobile Arm with Ratchet Knob]
Overall Length: 6.5" (16,5 cm)

3840-SA [Ratcheting Reduction
Stationary Arm]
Overall Length: 10.5" (26,7 cm)
Width: 9" (22,9 cm)
Height: 6" (15,2 cm)

INNOMED, INC

103 Estus Drive
Savannah, GA 31404
Tel 912.236.0000
Fax 912.236.7766

ISO 13485:2016

innomed.net
info@innomed.net

Clamp Kit

Kit includes:(1) Ratcheting Reduction
Stationary Arm, (1) Ratcheting
Reduction Mobile Arm with
Ratchet Knob (1) Plate Point,(1) Screw Point, and (2)Percutaneous Points

Innomed-Europe LLC

Alte Steinhauserstr. 19
CH-6330 Cham, Switzerland
Tel 0041 (0) 41 740 67 74
www.innomed-europe.com
orders@innomed-europe.com

High torque can help provide bone and joint reduction
without squeezing surrounding tissues
Swivel points are placed on the bone, plate, or screw
and the ratcheting dial is turned to the desired torque,
allowing hands free operation
Swivel point design allows the clamp to be easily
moved from x-ray view without losing reduction
Screw Point fits into a screw head
Plate Point fits into a 3.5 mm plate hole

Innomed-Europe GmbH

c/o Emons Logistik GmbHIn Rammelswiesen 9
D-78056 Villingen-Schwenningen, Germany
Tel 0049 (0) 7720 46110 60
www.innomed-europe.com
orders@innomed-europe.com

1.800.548.2362   INNOMED.NET

02
24© 2024 Innomed, Inc.
All Rights Reserved