202p

203p

203p

204p

205p

205p

206p

206p

207p

207p

MICROMETER
208p

208p

209p

209p

210p

210p

211p

211p

211p

212p

213p

213p

214p

214p

215p

215p

216p

216p

217p

218～219p

220p

221p

222p

222p

223p

223p

224p

224p

225p

225p

226p

226p

226p

三 校

04_201-226マイクロメータ.indd 201

CMYK

MICRO
METER
STAND

19/03/05 10:35:36

212p

MICRO
METER

2019/03/05 10:36

MICROMETER

S-LINE DIGITAL MICROMETER（IP67）

PDI
Importing measurement
data to PC▶

Reference standard
for calibration▶

S-line series micrometer, high end model for high performance.
IP67 that is suitable in wet environment.

Data Transfer Cable

Block Gauge

P42
〜49

P203

For holding micrometers▶
Micrometer Stand

Rust prevention▶
Zerust Film for Steel

P226

P380

MCD-30IP67S2

・IP rating for electrical equipment is defined based on IEC standard forwater and solid body protection・IP indication

IP 6 7

7

LiquidDegrees of protection against water
Definition
Testing forProtected against theeffects of temporary
immersion in water

Depth of 1m (height of theequipment is less than 850mm),
submerged for 30 minutes

USE●For measuring outer dimension
MATERIAL●Frame: Steel●Spindle: Stainless steel
FEATURES●Coolant proof●With IP67 rating, this product performs reliably even under harshconditions where it is exposed to a large amount of water, dust, or
cutting fluid●Work efficiency improves with 10mm/rev. fast displacement of themeasuring spindle●Wide measuring range, set of 3 micrometers will cover 0〜102mm
range●With 10mm/rev fast displacement of the measuring spindle●MCD-30IP67S2 has switchable measuring force（5N/10N）●Carbide tipped measuring face●Stand-by mode function（after approx. 20 min.）●Display hold function●One setting standard included（except for MCD-30IP67S2）POWER●POWER CR2032（lithium）battery included for testing● DIMENSIONS
D

56

C

F

Units: mm
Model No.

A

B

C

D

E

F

MCD-30IP67S2

37.3

27.5

68

12.5

63

181

MCD-66IP67S2

73.3

43

74

13.5

86

230

MCD-102IP67S2 109.3

60

74

14.5

106

269

Option: DATA TRANSFER CABLE⇒P203

19/03/18 16:26:46

A

φ28

MICRO
METER
STAND

Description

φ6.5

MICRO
METER

2nd digit

B

6

Degrees of protection against solid foreign objects
Description
Definition
Testing for
No ingress of dust
Complete protection against contact
SolidallowedE

1st digit

2nd digit: water proof rating
1st digit: direct‌contact-protection rating for ingress ofsolid foreign object※ Quantization error（± 1 digit）not included

Order No.

Model No.

Measuring Range（mm）

151227

MCD-30IP67S2

0 〜 30151228

MCD-66IP67S2

30 〜 66

151229MCD-102IP67S2

66 〜 102

Resolution
（mm）

0.001

Maximum Error ※（μm）

Weight

3

300g

4

440g

5

560g

202

念 校

04_201-226マイクロメータ.indd 202

CMYK

2019/03/18 16:26

PDI
Reference standard
for calibration▶

Blade design suitable for thin grooves. S-line series, high-end model for high performance.

Block Gauge

P42
〜49

MICROMETER

S-LINE DIGITAL BLADE MICROMETER（IP67）

Importing measurement
data to PC▶
Data Transfer Cable

P203
For holding micrometers▶
Micrometer Stand

P226
Rust prevention▶
Zerust Film for SteelP380● DIMENSIONSUnits: mm
56.4

68

6.5

φ28.5

34.3

φ6.5

63

15.5

27.5

USE
●For measuring narrow groove of round objects
MATERIAL●Frame: Steel●Spindle: Stainless steel/Measuring face: Carbide tipped
FEATURES●With IP67 rating, this products performs reliably even underharsh condition where it is exposed to a large amount of water,
dust or cutting fluid●Switchable measuring force（5N/10N）●Work efficiency improves with 10mm/rev fast displacement of
the measuring spindle●With 10mm/rev fast displacement of the measuring spindle●Stand-by mode（after approx. 20min.）●Display hold function
POWER●POWER CR2032 (lithium) battery included for testing

0.75

Option: DATA TRANSFER CABLE⇒P203

6.5

181

※ Quantization error（± 1 digit）not included

Order No.

Model No.

Measuring Range（mm）

Resolution（mm）

Maximum Error ※（μm）

Weight151234

MCD-25IP67BS2

0 〜 25

0.001

3

280g

DATA TRANSFER CABLE FOR S-LINE DIAL INDICATOR / VERNIER CALIPER / MICROMETER
Set of cable and software. Measured value can be imported to PC.
GO/NOGO judgment by tolerance set function
and recording measured value are possible.

（Cable）● Easy to know by color change and 
beep when out of tolerance● Measured value can be recorded automaticallyat a specific interval (0.1 ～ 60 min.)

USE●For importing measured value to PC
DTC-1
DTC-2
FEATURES●Measured value can be input directly to specified OS application such as Microsoft Excel or Notepad
COMPATIBILITY●Connectable product with DTC-1
152006 ～ 152008 S-LINE DIGITAL CALIPER（D-***IP67S）
151227 ～ 151229 S-LINE DIGITAL MICROMETER（MCD-**IP67S2）
S-LINE DIGITAL BLADE MICROMETER（MCD-25IP67BS2）
151234151726 ～ 151727
S-LINE DIGITAL INDICATOR（DEI-WKS*）
151716

MICRO
METER
MICRO
METER
STAND

P162

P202
P203

P241

151851 〜 151858 S-LINE DIGITAL THREE-POINT INTERNAL MICROMETER（MCD-****IP67S） P220
●Connectable product with DTC-2
151724 ～ 151725 S-LINE DIGITAL INDICATOR（DEI-233S*）

P240

151728 ～ 151729 S-LINE DIGITAL TEST INDICATOR（DET-234S*）

P242

Order No.

Model No.

152033DTC-1152034

DTC-2

Image of companion software
Downloading Android App [SLOG]

Connector type

Length

USB ※

3m ※

※ D-SUB9 pin type and cable in desired length are also available（USB type： Max. 7m / D-SUB9 pin type： Max. 10m）

203

念 校

04_201-226マイクロメータ.indd 203

CMYK

2019/03/18 17:10

19/03/18 17:10:40

APPLICABLE OS※●WindowsXP, Vista, 7, 8.1, 10 ●Android（Device with USB OTG support function）※Requires a USB host OTG cable
PRODUCT INCLUDES●Cable ●Software

MICROMETER

DIGITAL OUTSIDE MICROMETER（IP65）

PDI

IP65 that is suitable in wet environments

Reference standard
for calibration▶
Block Gauge

P42〜49
Zero point setting▶
Setting Standards

P225
For holding micrometers▶
Micrometer StandP226

MCD133-25IP

Rust prevention▶
Zerust Film for SteelP380

・IP rating for electrical equipment is defined based on IEC standard forwater and solid body protection・IP indication

IP 6 5

2nd digit:water proof rating
1st digit:direct contact-protection rating for ingress ofsolid foreign object

Degree of protection against solid foreign objects
Description
Definition
Testing for
Complete protection against
6
Solid
No ingress of dust allowed
contact
Degree of protection against water
2nd digit
Description
Definition
Testing forProtected against
Spray rotating equipment from 6.3mm
No harmful effects from direct
low pressure
nozzle, 12.5L/min flow from at distance of5
jet of water from any direction
water jet
2.5 to 3 m for more than three minutes
1st digit

MICRO
METER

● DIMENSIONS

MICRO
METER
STAND

USE
●For outer dimension measurement
MATERIAL●Frame: Ductile cast iron●Spindle: High carbon chrome steel
FEATURES●IP65 rating electronic digital micrometer, electronic is protectedagainst dust or water. Ideal to use for conditions where it is
exposed to water, dust, or cutting fluid.
●Carbide tipped measuring face●With a ratchet stop
●Auto display off function（after approx. 6 min.）●One setting standard included（except for MCD133-25IP）
POWER●One SR44（Silver oxide）battery included for testing

φC

B

φD

A

Units: mm
Model No.

A

B

MCD133-25IP

28

25

MCD133-50IP

52

32

MCD133-75IP

77

44

MCD133-100IP

103

56

C

D

6.5

18

19/03/18 16:26:59※ Quantization Error（± 1 digit）not included
Order No.

Model No.

Measuring Range（mm）

151196

MCD133-25IP

0 〜 25151197

MCD133-50IP

25 〜 50

151198

MCD133-75IP

50 〜 75

151199

MCD133-100IP

75 〜 100

Resolution（mm）

Maximum Error ※
（μm）

4
0.001
6

Weight

275g
380g
460g
590g

204

念 校

04_201-226マイクロメータ.indd 204

CMYK

2019/03/18 16:27

PDI

Great value for the performance

Reference standard
for calibration▶
Block Gauge

P42
〜49

MICROMETER

DIGITAL OUTSIDE MICROMETER

Zero point setting▶
Setting Standards

MCD130-25

P225
For holding micrometers▶
Micrometer Stand

P226
Rust prevention▶
Zerust Film for SteelMCD130-125

P380● DIMENSIONS/&'',63/$<

MCD130-25〜MCD130-100

φC

φD

A

B

LCD DISPLAY

MICRO
METER
MICRO
METER
STAND

MCD130-125,MCD130-150

USE●Measuring outer dimension
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●Carbide measuring faces●With ratchet stop
●Auto display off function（after approx. 6 min.）●With thermal protection（except for MCD130-125, MCD130-150）●Painted frame●Sphere spindle attachment（SR2.5）●One setting standard included（except for MCD130-25）
POWER●One SR44（Silver oxide）battery included for testing
Order No.

Model No.

Measuring Range（mm）

MCD130-25

0 〜 25

MCD130-50

25 〜 50

151263

MCD130-75

50 〜 75

151264

MCD130-100

75 〜 100

151266

MCD130-125

100 〜 125

151267

MCD130-150

125 〜 150

Model No.

A

B

MCD130-252824MCD130-505331

MCD130-757843MCD130-100

103

51

MCD130-125

129

75

MCD130-150

154

85

デジタル外側マイクロメータ
T_326

C

D

6.5

18

MCD130-125※ Quantization Error（± 1 digit）not included
Resolution（mm）

Maximum Error ※（μm）

4

Weight

260g
380g
470g

0.001
6

640g
490g
570g

205

念 校

04_201-226マイクロメータ.indd 205

CMYK

2019/03/18 16:27

19/03/18 16:27:03

151261151262

Units: mm

MICROMETEROUTSIDE MICROMETER

MC105 / MC106 SERIESPDI
Reference standard
for calibration▶

Great value for the performance.
For more stable measurement compared to vernier calipers.

Zero point setting▶
Setting Standards

Block Gauge

P42
〜49

P225

For holding micrometers▶
Micrometer Stand

Rust prevention▶
Zerust Film for Steel

P226

P380

MC105-25● DIMENSIONS

MC105 series

MC106-175

MC106 series

Units: mm
Model No.

MICRO
METER

19/03/05 10:36:37

MICRO
METER
STAND

USE
●Measuring outer dimension
MATERIAL●Frame: Carbon steel for machine use●Spindle: High carbon chrome steel
FEATURES●Carbide measuring face
●With ratchet stop for constant pressure●Painted frame●One setting standard included（except for MC105-25）
Order No.

Model No.

Measuring Range（mm）

151070MC105-25

0 〜 25151071

MC105-50

25 〜 50

151072

MC105-75

50 〜 75

151073

MC105-100

75 〜 100

151074

MC105-125

100 〜 125

151075

MC105-150

125 〜 150

151401MC106-175

150 〜 175

151402

MC106-200

175 〜 200

151403

MC106-225

200 〜 225

151404

MC106-250

225 〜 250

151405

MC106-275

250 〜 275

151406

MC106-300

275 〜 300

A

B

MC105-25

31

29

MC105-50

56

37

MC105-75

81

50

MC105-100

106

63

MC105-125

131

75

MC105-150

156

88

MC106-175

183

104

MC106-200

209

117

MC106-225

234

130

MC106-250

260

142

MC106-275

285

155

MC106-300

310

168

Resolution（mm）

Maximum Error
（μm）

4
5
6
0.01
7
8
9

C

D

6.5

18

Weight

150g
200g
230g
260g
330g
400g
670g
740g
870g
900g
970g
1070g

206

三 校

04_201-226マイクロメータ.indd 206

CMYK

2019/03/05 10:36

PDI
Reference standard
for calibration▶

A set of six different sizes of outside micrometers in a wooden box

Block Gauge● Refer to MC105 series for specifications

Zero point setting▶
Setting Standards

P42
〜49

P225

For holding micrometers▶
Micrometer Stand

MICROMETEROUTSIDE MICROMETER SET

Rust prevention▶
Zerust Film for Steel

P226

P380

● Included in set
OutsideMicrometers
SettingStandards

0〜25,25〜50,50〜75,
75〜100,100〜125,
125〜150（6 pcs）
25,50,75,100,125（5 pcs）

Order No.

Model No.

Measuring Range（mm）

Resolution（mm）

Weight151240MC105-150ST

0 〜 150

0.01

3.8kg

OUTSIDE MICROMETER WITH ADJUSTABLE ANVIL

PDI

Wide measuring range with adjustable anvil

Reference standard
for calibration▶
Block Gauge

P42〜49● DIMENSIONS$19,/$'-867$%/(

MICRO
METER

MC104-400

MICRO
METER
STAND

USE
●Measuring outer dimension
MATERIAL●Spindle: High carbon chrome steel
FEATURES●Changing mounting position of anbil by adjustable anvil forwide mesuring range●Lightweight flame structure●With ratchet stop
●Painted frame●Carbide measuring faces
SPECIFICATIONS●Adjustable anvil：25mm 1pc / 50mm 1pc included●Head stroke：25mm
Model No.

Measuring Range（mm）

151280MC104-400

151281
151282

Resolution（mm）Model No.

A

B

MC104-400

409

224

MC104-500

509

275

MC104-600

609

321

MC104-700

709

371

MC104-800

809

426

MC104-900

909

476

MC104-1000

1009

576

C

D

8

18

Maximum Error（μm）

Setting Standard

Weight300 〜 400

11

325 & 375（mm）1EA

5.5kg

MC104-500

400 〜 50013425 & 475（mm）1EA

6kg

MC104-600

500 〜 600

15525 & 575（mm）1EA

6.5kg

151283

MC104-700

600 〜 70016625 & 675（mm）1EA

10kg

151284

MC104-800

700 〜 80018

725 & 775（mm）1EA

15kg

151285

MC104-900

800 〜 900

20

825 & 875（mm）1EA

18kg

151286

MC104-1000

900 〜 1000

20

925 & 975（mm）1EA

20.5kg

0.01

207

念 校

04_201-226マイクロメータ.indd 207

CMYK

2019/03/18 16:27

19/03/18 16:27:07

Order No.

Units: mm

MICROMETER

DIGITAL MICROMETER WITH INTERCHANGEABLE ANVIL

PDI

Wide measuring range with interchangeable anvils

Reference standard
for calibration▶
Block Gauge

For holding micrometers▶
Micrometer Stand

P42
〜49

P226
Rust prevention▶
Zerust Film for SteelMCD136-100

USE●Measuring outer dimension
MATERIAL●Frame: Carbon steel ●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●One for wide measuring range by interchangeable anvils●Carbide measuring faces ●With ratchet stop
●Auto display off function（after approx. 5 min.） ●Painted frame
SPECIFICATIONS●Interchangeable anvils: 4 pcs●Setting standard：MCD136-100：3 pcs（25・50・75mm）：MCD136-200：4 pcs（100・125・150・175mm）●Head stroke: 25mm
POWER
●One SR44（Silver oxide）battery included for testing
Order No.

Model No.

Measuring Range（mm）

152151MCD136-100

0 〜 100

152152

MCD136-200

100 〜 200

P380

Interchangeable anvils● DIMENSIONS/&'',63/$<

Units: mm
Model No.

A

B

MCD136-100

108

57

MCD136-200

208

114

C

D

8

18

※ Quantization Error（± 1 digit）not included
Resolution（mm）

Maximum Error ※
（μm）

0.001

Weight

6

500g

8

940g

MICROMETER WITH INTERCHANGEABLE ANVIL

PDI
Reference standard
for calibration▶

Wide measuring range with interchangeable anvils

Block Gauge

P42
〜49

MICRO
METER

For holding micrometers▶

MC1064-100

MICRO
METER
STAND

Micrometer Stand

P226
Rust prevention▶
Zerust Film for Steel

P380

19/03/18 16:27:12

Interchangeable anvils

USE●Measuring outer dimension
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●One for wide measuring range by interchangeable anvils●Carbide measuring faces●With ratchet stop
●Painted frame
SPECIFICATIONS●Interchangeable anvil: 4 pcs●Setting standard：MC1064-100：3 pcs（25・50・75mm）：MC1064-200：4 pcs（100・125・150・175mm）●Head stroke: 25mm
Order No.

Model No.

Measuring Range（mm）

151441MC1064-100

0 〜 100

151442

MC1064-200

100 〜 200

● DIMENSIONSUnits: mm
Model No.

A

B

MC1064-100

107

59

MC1064-200

208

116

Resolution
（mm）

0.01

C

D

8

18

Maximum Error（μm）

Weight

5

380g

7

810g

208

念 校

04_201-226マイクロメータ.indd 208

CMYK

2019/03/18 16:27

PDI

Deep throat frame for measuring thickness of sheet metal and so on

Reference standard
for calibration▶
Block Gauge

P42〜49
MCD233-100U

MICROMETER

DIGITAL DEEP THROAT MICROMETER● DIMENSIONS/&'',63/$<

USE●Deep throat frame for measuring thickness of sheet metal,
paper and sheet plastic
MATERIAL●Frame: Carbon steel ●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value ●Carbide measuring faces●With ratchet stop ●With thermal protection cover●Auto display off function（after approx. 5 min.） ●Painted frame
POWER●One SR44（Silver oxide）battery included for testing)/$7)/$7

63+(5,&$/)/$7

Units: mm
Model No.

A

MCD233-150U

B

C

D

6.5

18

100

MCD233-100U
32

MCD233-300U

150
300

※ Quantization Error（± 1 digit）not included
Order No.

Model No.

152111MCD233-100U

152112

MCD233-150U

152113

MCD233-300U

Measuring Range（mm）

0 〜 25Resolution（mm）

Maximum Error ※（μm）

Shape ofMeasuring Face

6

Flat-Flat

8

Spherical-Flat

0.001

Weight

720g
960g
2000g

DEEP THROAT MICROMETER

PDI
Reference standard
for calibration▶

Deep throat frame for measuring thickness of sheet metal and so on

Block Gauge

P42
〜49

MICRO
METER
STAND

MC203-100U● DIMENSIONSUSE●Deep throat frame for measuring thicknessof sheet metal, paper and sheet plastic
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Carbide measuring faces●With ratchet stop
●With thermal protection cover●Painted frame
Model No.

151451

MC203-100U

151452

MC203-150U

151453

MC203-300U)/$7)/$7

63+(5,&$/)/$7

Units: mm
Model No.

A

MC203-100U
MC203-150U
MC203-300U

Measuring Range（mm）

0 〜 25Resolution
（mm）

0.01

B

C

D

6.5

18

100
31

150
300

Maximum Error（μm）

Shape ofMeasuring Face

4

Flat-Flat

5

Spherical-Flat

Weight

580g
820g
1980g

209

念 校

04_201-226マイクロメータ.indd 209

CMYK

2019/03/18 16:27

19/03/18 16:27:16

Order No.

MICRO
METER

MICROMETER

DIGITAL SMALL MEASURING FACE MICROMETERPDISuitable for groove measurement on splined shaft by small measuring faces
Reference standard
for calibration▶

For holding micrometers▶
Micrometer Stand

Block Gauge

MCD230-25SA

USE●Small measuring faces for measuring splined shafts,
slots and keyways
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●Carbide measuring faces●With ratchet stop
●With thermal protection cover●Auto display off function（after approx. 5 min.）●Painted frame
POWER●One SR44（Silver oxide）battery included for testing
Order No.

Model No.

152121MCD230-25SA

152122

MCD230-25SB

P42
〜49

Rust prevention▶
Zerust Film for Steel

P226

P380

● DIMENSIONSMCD230-25SA

MCD230-25SB/&'',63/$<

Units: mm
Model No.
MCD230-25SA
MCD230-25SB

A

B

C

D

57

32

6.5

18

※ Quantization Error（± 1 digit）not included

Measuring Range（mm）

Resolution（mm）

Maximum Error ※（μm）

0 〜 250.001

4Measuring Face（mm）

Weight

φ2×5370g

φ 3 × 10380g

SMALL MEASURING FACE MICROMETERPDISuitable for groove measurement on splined shaft by small measuring faces
Reference standard
for calibration▶

For holding micrometers▶
Micrometer Stand

Block Gauge

MICRO
METER

P42
〜49

MICRO
METER
STAND

P226

P380

● DIMENSIONS
MC200-25SA

MC200-25SA

19/03/18 16:27:21

Rust prevention▶
Zerust Film for Steel

USE●Small measuring faces for measuring splined shafts,
slots and keyways
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Carbide measuring faces●With ratchet stop
●With thermal protection cover●Painted frame
Order No.

Model No.

151461MC200-25SA

151462MC200-25SB

MC200-25SB

Units: mm
Model No.
MC200-25SA
MC200-25SB

Measuring Range（mm）

Resolution（mm）

Maximum Error（μm）

0 〜 25

0.01

4

A

B

C

D

56

31

6.5

18

Measuring Face
（mm）

φ2×5
φ 3 × 10Weight

320g

210

念 校

04_201-226マイクロメータ.indd 210

CMYK

2019/03/18 16:27

PDI

Suitable for thin groove measurement by blade type measuring faces
Reference standard
for calibration▶
Block Gauge

Zero point setting▶

P42
〜49

MCD235-25BA

For holding micrometers▶

Setting Standards

Rust prevention▶

Micrometer Stand

P225

Zerust Film for SteelP226

MICROMETER

DIGITAL BLADE MICROMETER

P380● DIMENSIONS
6

6

0.75

USE
●Non-rotating spindle, with blade type measuring faces
MATERIAL●Frame: Carbon steel※Above dimension is for BA type●Spindle: High carbon chrome steel
FEATURES
Model No.
A
●Quick digital display of the measured value
MCD235-25BA
57●With ratchet stop
●With thermal protection cover
MCD235-50BA
82●Auto display off function（after approx. 5 min.）●Painted frame●One setting standard included（MCD235-50BA, MCD235-50BAU）
MCD235-25BAU
57
POWER
MCD235-50BAU
82●One SR44（Silver oxide）battery included for testing

6.5

0.4

BA type

3.5

BAU type

Units: mm
B

C

32

D
直進式ブレードマイクロメータ
（細幅）

18
φC

32

No.13

A

8

φD

44.5

B

44.5

※ Quantization Error（± 1 digit）not included

Order No.

Model No.

Measuring Range（mm）

152131MCD235-25BA

0 〜 25152132

MCD235-50BA

25 〜 50

152133

MCD235-25BAU

0 〜 25152134

MCD235-50BAU

25 〜 50

Maximum Error ※（μm）

Resolution（mm）

Weight

400g
0.001

530g

4

400g
530g

BLADE MICROMETERPDISuitable for thin groove measurement by blade type measuring faces
Reference standard
for calibration▶
Block Gauge

Zero point setting▶
Setting Standards

P42
〜49

MC201-25BA

For holding micrometers▶
Micrometer Stand

P225

Rust prevention▶
Zerust Film for Steel

P226

P380

● DIMENSIONS

MICRO
METER
STAND

MC201-25BB

6

0.4

BA type
USE※Dimension above is BA type●Non-rotating spindle, with blade typemeasuring faces
Model No.
A
MATERIAL
MC201-25BA
56●Frame: Carbon steel
MC201-50BA
81●Spindle: High carbon chrome steel
FEATURES
MC201-25BB
56●With ratchet stop
MC201-50BB
81●With thermal protection coverMC201-25BAU
56●Painted frame●One setting standard included（MC201-50BA, MC201-50BB, MC201-50BAU） MC201-50BAU
81
Order No.

Model No.

Measuring Range
（mm）

MICRO
METER

Resolution（mm）

Maximum Error（μm）

BB typeBAU type
Units: mm

B

C

D

31
44
31
44

直進式ブレードマイクロメータ
（細幅）
No.13

8

18

31
44
Weight

MC201-25BA

0 〜 25320g

MC201-50BA

25 〜 50410g

151251MC201-25BB

0 〜 25151252MC201-50BB

25 〜 50

151473

MC201-25BAU

0 ～ 25280g

151474

MC201-50BAU

25 ～ 50

380g

4

320g
410g

211

念 校

04_201-226マイクロメータ.indd 211

CMYK

2019/03/18 16:27

19/03/18 16:27:25

1514711514720.01

3.5

MICROMETER

DIGITAL GEAR TOOTH MICROMETERPDISuitable for thickness measurement of gear and paper by disk type measuring faces
Reference standard
for calibration▶

Zero point setting▶

P42〜49
MCD230-25D

For holding micrometers▶

Setting Standards

Block Gauge

Micrometer Stand

P225

Rust prevention▶
Zerust Film for Steel

P226

P380

● DIMENSIONS

5(/,()USE●Disk type measuring faces for measuring gear and paper thickness
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●With ratchet stop●With thermal protection cover●Auto display off function（after approx. 5 min.）●Painted frame●One setting standard included（MCD230-50D）
POWER●One SR44（Silver oxide）battery included for testing/&'',63/$<

Units: mm
Model No.

A

B

MCD230-25D

43

32

MCD230-50D

68

44.5

C

D

E

F

G

8

18

8

20

0.7

※ Quantization Error（± 1 digit）not included
Order No.

Model No.

Measuring Range（mm）

152141MCD230-25D

0 〜 25152142

MCD230-50D

25 〜 50

Resolution（mm）

Maximum Error ※（μm）

Gear Module

0.001

4

0.5 〜 6Weight

400g
540g

GEAR TOOTH MICROMETERPDISuitable for thickness measurement of gear and paper by disk type measuring faces
Reference standard
for calibration▶

Zero point setting▶

MICRO
METER

P42
〜49

MICRO
METER
STAND

For holding micrometers▶

Setting Standards

Block Gauge

Micrometer Stand

P225

Rust prevention▶
Zerust Film for Steel

P226

P380

● DIMENSIONS

MC200-25D

19/03/18 16:27:30

5(/,()USE●Disk type measuring faces for measuring
gear and paper thickness
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●With ratchet stop
●With thermal protection cover●Painted frame●One setting standard included（MC200-50D）
Measuring Range（mm）

Order No.

Model No.

151481MC200-25D

0 〜 25151482MC200-50D

25 〜 50

Units: mm
Model No.

A

B

MC200-25D

34

22

MC200-50D

59

31

C

D

E

F

G

8

18

8

20

0.7

Resolution（mm）

Maximum Error（μm）

Gear Module

0.01

4

0.5 〜 6Weight

240g
330g

212

念 校

04_201-226マイクロメータ.indd 212

CMYK

2019/03/18 16:27

PDI

Suitable for wall thickness mesurement of tubes by a spherical measuring face

Reference standard
for calibration▶
Block Gauge

P42
〜49

MICROMETER

DIGITAL TUBE MICROMETER

For holding micrometers▶
Micrometer Stand

P226
Rust prevention▶
Zerust Film for Steel

P380

USE
●For measuring wall thickness of various tubes
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●Carbide measuring faces●With ratchet stop
●With thermal protection cover●Auto display off function（after approx. 5 min.）●Painted frame
POWER●One SR44（Silver oxide）battery
included for testing● DIMENSIONSUnits: mm/&'',63/$<

Can be used for tubesof IDφ10mm and larger※ Quantization Error（± 1 digit）not included
Order No.

Model No.

Measuring Range（mm）

152161MCD236-25T

0 〜 25Resolution（mm）

Maximum Error ※（μm）

Weight

0.001

4

270g

TUBE MICROMETER

PDI
Reference standard
for calibration▶

Suitable for wall thickness mesurement of tubes by a spherical measuring face

Block Gauge

P42
〜49

MICRO
METER
MICRO
METER
STAND

For holding micrometers▶
Micrometer Stand

P226
Rust prevention▶
Zerust Film for SteelP380● DIMENSIONSCan be used for tubesof IDφ10mm and larger

Order No.

Model No.

Measuring Range（mm）

Resolution（mm）

Maximum Error（μm）

Weight151436MC200-25T

0 〜 25

0.01

4

200g

213

念 校

04_201-226マイクロメータ.indd 213

CMYK

2019/03/18 16:27

19/03/18 16:27:35

USE●For measuring wall thickness of various tubes
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Carbide measuring faces●With ratchet stop
●With thermal protection cover●Painted frame

Units: mm

MICROMETER

DIGITAL POINT MICROMETERPDISuitable for measuring drills, grooves and hard-to-reach dimensions
by pointed measuring faces

Reference standard
for calibration▶

For holding micrometers▶
Micrometer Stand

Block Gauge

P42
〜49

P226
Rust prevention▶
Zerust Film for Steel

P380

USE
●Pointed spindle and anvil for measuring
drills, grooves and hard-to-reach dimensions
MATERIAL●Frame: Carbon steel ●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●With ratchet stop ●Carbide measuring faces●With thermal protection cover●Auto display off function（after approx. 5 min.） ●Painted frame
SPECIFICATIONS●Point: 15°
POWER●One SR44（Silver oxide）battery included for testing

MICRO
METER

19/03/18 16:27:38

MICRO
METER
STAND

Order No.

Model No.

Measuring Range（mm）

152171MCD232-25P

0 〜 25● DIMENSIONSUnits: mm

7,3

7,3

/&'',63/$<※ Quantization Error（± 1 digit）not included
Resolution（mm）

Maximum Error ※/&'',63/$<（μm）

Weight

0.001

4

350g

POINT MICROMETER

PDI

standard
Suitable for measuring drills, grooves and hard-to-reach dimensions Reference
for calibration▶
Block Gaugeby pointed measuring faces

For holding micrometers▶
Micrometer Stand

P42
〜49

USE
●Pointed spindle and anvil for measuring drills, grooves
and hard-to-reach dimensions
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●With ratchet stop ●Carbide measuring faces●With thermal protection cover ●Painted frame
SPECIFICATIONS●Point: 15°

Rust prevention▶
Zerust Film for SteelP226● DIMENSIONS

P380

Units: mm

7,3

Order No.

Model No.

Measuring Range（mm）

Resolution（mm）

Maximum Error（μm）

Weight151431MC200-25P

0 〜 25

0.01

4

320g

214

念 校

04_201-226マイクロメータ.indd 214

CMYK

2019/03/18 16:27

PDI

Measured value can be read on both front and back sides.
Suitable for measuring in narrow space on thimble side and for the left-handed.

Reference standard
for calibration▶

For holding micrometers▶
Micrometer Stand

Block Gauge

P42
〜49

P226

MICROMETERLR TYPE MICROMETER

Rust prevention▶
Zerust Film for SteelP380● DIMENSIONSUnits: mm
31

24

φ6.5

φ18

USE
●Measuring outer dimension
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Carbide measuring faces●With thermal protection cover●Can be read on either side
Order No.

Model No.

Measuring Range（mm）

Resolution（mm）

Maximum Error（μm）

Weight151026MC101-25LR

0 ～ 25

0.01

4

200g

OUTSIDE COUNT MICROMETER

LRマイクロメータ

PDI

T_325

Mechanical digit counter for misreading prevention

Reference standard
for calibration▶

Zero point setting▶
Setting Standards

Block Gauge

P42
〜49

P225

For holding micrometers▶
Micrometer Stand

Rust prevention▶
Zerust Film for Steel

P226

P380

MICRO
METER
MICRO
METER
STAND

MC122-25C● DIMENSIONSUSE●Measuring outer dimension
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Mechanical digit counter for quick and error-freereading●Carbide measuring faces●With ratchet stop
●Painted frame●One setting standard included（MC122-50C）&2817(5Units: mm

Model No.

151391MC122-25C

0 〜 25151392

MC122-50C

25 〜 50A

B

MC122-25C

28

25

MC122-50C

53

32

Resolution（mm）

Maximum Error
（μm）

0.01

4

C

D

6.5

18

Weight

220g
300g

215

念 校

04_201-226マイクロメータ.indd 215

CMYK

2019/03/18 16:27

19/03/18 16:27:43

Measuring Range（mm）

Order No.

Model No.

MICROMETER

HUB MICROMETERPDISuitable for measuring in narrow place where regular micrometer does not fit

Reference standard
for calibration▶
Block Gauge

Zero point setting▶
Setting Standards

P42
〜49

P225
Rust prevention▶
Zerust Film for SteelP380
MC203-25H● DIMENSIONS

USE●Measuring hub thickness and shoulders inside bore
MATERIAL●Frame: Carbon steel ●Spindle: High carbon chrome steel
FEATURES●With ratchet stop ●Painted frame●Carbide measuring faces●One setting standard included（MC203-50H）
Order No.

Model No.

Measuring Range（mm）

151381

MC203-25H

0 〜 25151382

MC203-50H

25 〜 50

Units: mm
Model No.

A

B

MC203-25H

28

20

MC203-50H

53

22

Resolution（mm）

Maximum Error
（μm）

0.01

4

C

D

E

6

6.5

18

Weight

150g
170g

MICROMETER WITH DIAL GAUGE
MICRO
METER

PDI

For direct GO/NG judgment of mass-produced parts attaching on micrometer stand

Reference standard
for calibration▶
Block Gauge

P42
〜49

MICRO
METER
STAND

For holding micrometers▶
Micrometer Stand

Rust prevention▶
Zerust Film for Steel

P226

P380

Units: mm,1',&$725/2&..12%$19,/5$1*(㸸



ȭ

ȭ

USE
●For direct GO/NOGO judgment of
mass-produced parts
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel●Dial gauge：SteelFEATURES●Enables comparative measurement by setting standard value
with the spindle lock and read the dial gauge as running parts,1',&$72502817',$0(7(5ȭ㸩



ȭ

19/03/18 16:27:48● DIMENSIONS/(9(5,1',&$725',$/*$8*(PP,1&/8'('

Order No.

Model No.

Measuring Range（mm）

Resolution（mm）Micrometer Head
Maximum Error（μm）

Dial Gauge Error overentire range（μm）

Weight151490

MC204-25DG

Micrometer Head： 0〜25
Dial Gauge： 0〜10

0.01

4

± 14

545g

216

念 校

04_201-226マイクロメータ.indd 216

ࢲ࢖ࣖࣝࢤ࣮ࢪ௜࣐࢖ࢡ࣓࣮ࣟࢱࠉ0&'*
7B

CMYK

2019/03/18 16:27

PDI

For precision measurement of mass-produced parts attaching on micrometer stand

Reference standard
for calibration▶
Block Gauge

P42〜49
For holding micrometers▶

MICROMETER

INDICATING MICROMETER

Rust prevention▶

Micrometer Stand

Zerust Film for Steel

P226

P380

● DIMENSIONSUnits: mm/,0,7,1',&$725

386+%87721

7+(50$/3527(&7,21&29(5① Select a gauge standard (master)
Example- 5.000mm Ceramic gauge block②Turn the thimble to set the scale to read
5.00 and then lock the spindle with spindle lockSpindle lock

③While pressing release button insert
gauge block and then release button tocatch block in spindle
Gauge Block

Thimble④ Rotate the dial scale to align zerowith pointerPointer② insert① press

RotatePush button

Scale plate⑤ Press the button again and remove
the gauge block. Micrometer is readyfor measurement.

MICRO
METER

※Using the limit indicators, you can
instantly determine if a part is withintolerance

MICRO
METER
STAND

Limit indicator
Gauge Block② Remove
Tolerance range① Press

Push button

Order No.

Model No.

152101

MC263-25IS

Measuring Range（mm）

Resolution（mm）

Maximum Error（μm）

Micrometer Head： 0〜25

Micrometer Head：0.002

Micrometer Head：3

Dial Indicator： ±0.04

Dial Indicator：0.001

Dial Indicator：1

Weight

530g

217

念 校

04_201-226マイクロメータ.indd 217

CMYK

2019/03/18 16:27

19/03/18 16:27:51

USE●Retractable anvils with an indicator for precision
measurement of mass-produced parts
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Carbide tipped measuring faces●With Thermal protection cover●Dial indicator has limit indicator clips

MICROMETER

ELECTRIC MICROMETER PROBENEW!

For high precision measuring of 0.01μm resolution by using
SNAP GAUGE and ELECTRIC MICROMETER DISPLAY

Display▶
Micrometer Display

P219
Measuring Instruments▶
S-Line Snap Gauge

P219

To be used with ELECTRIC MICROMETER DISPLAY M3 and SNAP GAUGE (sold separately). This unit cannot be functioned
without them.

MICRO
METER
MICRO
METER
STAND

19/03/05 10:37:39

EM-PHTL

Order No.

Model No.

Type

NEW! 152092

EM-PHTS

Straight

NEW! 152093

EM-PHTL

L shape● DIMENSIONSUnits: mm









࠙(03+7/ࠚ







ȭ

࠙(03+76ࠚȭ

USE●For high precision measurement
MATERIAL●SteelFEATURES●Applicable to 0.01um resolution by using ball bearing●To be combined with ELECTRIC MICROMETER DISPLAY M3 and
SNAP GAUGE（sold separately）
SPECIFICATIONS●Linearity: 0.25%/ ±1mm
●Measuring force: ①0.63N ②0.16～4.0N（by spring conversion)●Outer mount diameter: φ8mm - Fit tolerance h6●Carrier frequency: 13kHz●Input voltage: 3.0V
●Durability: Approx. 10,000,000 times●Cable length: 2m
POWER●Supplied from ELECTRIC MICROMETER DISPLAY M3

ȭ

EM-PHTS

Measuring Range（mm）

Movement Range（mm）

Resolution（μm）

Repeat error
（μm）

±1

4.6

0.01

0.01

Weight

120g
120g

218

三 校

04_201-226マイクロメータ.indd 218

CMYK

2019/03/05 10:37

NEW!

High reliability 0.01μm resolution.
For displaying measurement value by combination use of ELECTRIC MICROMETER PROBE (sold separately).
USE●For displaying measurement value by combination use of ELECTRIC
MICROMETER PROBE EM-PHTS / EM-PHTL
MATERIAL ●Body: AluminumFEATURES●Simultaneou display of up to two channels. Tolerance of each channel
can be set independently. ●Measurement data can be output to PC●Touch-screen display ●Color display. GO/NOGO judgement and control
point are shown in three colors (green, red, yellow).
●Display mode can be chosen from dial display or bar-graph display
SPECIFICATIONS●two channels simultaneous display●Math function (Max. value, Min. value, Max. (minus) Min. value, average value and median value)●Independent tolerance setting
POWER ●DC5V (AC adaptor included)● DIMENSIONS Units: mm





Order No.

Model No.

Resolution（μm）

Output Signal

Applicable Models

WeightNEW! 152091

M3

0.01

RS232C, USB

EM-PHTS
EM-PHTL

600g

S-LINE SNAP GAUGE AND SNAP GAUGE STAND

MICROMETER

ELECTRIC MICROMETER DISPLAY

NEW!
Reference standard
for calibration▶

For high precision inspection by using ELECTRIC
MICROMETER PROBE and ELECTRIC MICROMETERDISPLAY, and digital indicator (each sold separately)

Block Gauge・Pin Gauge

P42
〜49

ALD-040

Usage Example

Measuring instruments▶
Electronic Micrometer

ALD-4080

P218〜219
Measuring instruments▶
Digital Indicator

P240〜241
・P243
Rust prevention▶
Zerust Film for Steel

ALD-080

USE●Outer measurement
MATERIAL
SNAP GAUGE STAND●Body: Steel ●Probe sensor: Carbide steel
ASD-1
FEATURES●Applicable to optional equipment（ELECTRIC MICROMETER PROBE EM-PHTS and EM-PHTL andindicators）●Applicable to size φ8mm sensor（for electronic micrometer and linear sensor）●Optional stand ASD-1（sold separately）
SPECIFICATIONS●Parallelism: Under 1μm ●Flatness: Under 0.2μm●Repeat accuracy: Under 0.5μm ●Sensor mount: φ8mm
Order No.

Measuring range（mm）

Model No.

ALD-040

0 〜 35

ALD-4080

35 〜 75

NEW! 151903

ALD-080

0 〜 70NEW! 151904

ALD-80150

75 〜 140

MICRO
METER
STAND

ALD-80150

Probe sensor movement range（mm）

Measurement unit（mm）

φ 2 L5.5 (each bar)
6

φ9
φ 11

Weight

380g
630g
650g
1.78kg■ SNAP GAUGE STAND
Order No.

Model No.

Material

Base Size（mm）

Applicable Models

WeightNEW! 151900

ASD-1

Steel

80 × 120

ALD-040, ALD-4080, ALD-080, ALD-80150

1.3kg

219

三 校

04_201-226マイクロメータ.indd 219

CMYK

2019/03/05 10:37

19/03/05 10:37:46

NEW! 151901
NEW! 151902

P380

MICRO
METER

MICROMETER

S-LINE DIGITAL THREE-POINT INTERNAL MICROMETER

PDI
Reference standard
for calibration▶

S-line series, high-end model for high performance.
IP67 that is suitable in wet environment.

Ring Gauge

Importing measurement
data to PC▶
Data Transfer Cable

P68
〜72

Rust prevention▶
Zerust Film for Steel

P203

P380

MCD-0810IP67S・IP rating for electrical equipment is defined based on IEC60529
standard for water・IP code

1st digit
6
2nd digit
7

MICRO
METER
MICRO
METER
STAND

2nd digit: water proof rating
1st digit: direct contact-protection rating for ingress ofsolid foreign object

Degrees of protection against solid foreign objects
Description
Definition
Testing for
No ingress of dust
Complete protection against contact
SolidallowedDegrees of protection against water
Description
Definition
Testing for
LiquidProtected against theeffects of temporary
immersion in water

Depth of 1m (height of theequipment is less than 850mm),
submerged for 30 minutes

USE●Measurement of internal diameter
MATERIAL●Gauge head: Alloy tool steel（for MCD-10125IP67S and smaller sizes）
Carbide tipped（for MCD12516IP67S and larger sizes）
FEATURES●IP67 rating digital micrometer. Electronic unit is protected against dust
and water.
●Four preset values can be set●Measurement at the very bottom is possible as less than 0.1mm between
probes and the edge（MCD-12516IP67S and larger size）●With a ratchet stop
●Auto display off function（after approx. 4 hrs.）
POWER●CR2032（lithium）battery included for testing● DIMENSIONS



#
$
%

MODE

SET

&

Units: mm
Model No.
MCD-0608IP67S
MCD-0810IP67S
MCD-10125IP67S
MCD-12516IP67S
MCD-1620IP67S
MCD-2025IP67S
MCD-2535IP67S
MCD-3550IP67S

Option: DATA TRANSFER CABLE⇒P203

Units: mm

IP 6 7

A

B

C

D

166

60

less than 1.7

167

62

less than 2

168

63

166

67

179

80

2.8
3
4
6.3
8.3

less than 0.1

11.4
13
18.3

19/03/18 16:27:57※ Quantization Error（± 1 digit）not included
Measuring Range（mm）

Resolution（mm）

Maximum Error ※（μm）

Order No.

Model No.

151851

MCD-0608IP67S

6

〜 8

220g

151852

MCD-0810IP67S

8

〜 10

230g

〜 12.5245g

Weight151853

MCD-10125IP67S

10151854

MCD-12516IP67S

12.5 〜 16

151855MCD-1620IP67S

16

〜 20

151856

MCD-2025IP67S

20

〜 25

345g

151857

MCD-2535IP67S

25

〜 35

365g

151858MCD-3550IP67S

35

〜 50

505g

0.001

4

270g
285g

＊ Three point micrometer is precision made for use in a condition of around 20℃ for the best performance. Under extreme temperature
conditions, measurement error or operation of the instrument may be affected.
＊ Ring gauge calibration standard is not included. Separate purchase is required.

220

念 校

04_201-226マイクロメータ.indd 220

CMYK

2019/03/18 16:28

PDI
Reference standard
for calibration▶

S-line series, high-end model for high performance

Ring Gauge

Rust prevention▶
Zerust Film for Steel

P68
〜72

P380

MICROMETER

S-LINE THREE-POINT INTERNAL MICROMETER

MCA-10125S● DIMENSIONS
#
Ǿ

$
%

&

MCA-0608S 〜 MCA-1620S

$

ȭ

%
&
'
MC-2025IPS 〜 MC-3550IPS
Units: mm
Model No.
MCA-0608S

USE●Measurement of internal diameter
MATERIAL●Gauge head：Alloy tool steel（for MC-10125IPS and smaller sizes）
Carbide tipped（for MC-12516IPS and larger sizes）
FEATURES●Measuring at the very bottom is possible as distance between
probes and the edge is short（MCD-12516IP67S and larger size）●With a ratchet stop
Measuring Range（mm）

Order No.

Model No.

151761

MCA-0608S

151762

MCA-0810S

8〜 10151763

MCA-10125S

10〜 12.5151764

MCA-12516S

12.5 〜 16

151765

MCA-1620S

16

6

MCA-0810S
MCA-10125S
MCA-12516S
MCA-1620S

A

B

C

121

60

less than 1.7

123

62

less than 2

125

63

MC-2025IPS

148

MC-2535IPS

150

MC-3550IPS

163

Resolution
（mm）

74

Maximum Error
（μm）

20

〜 25

25

〜 35

151420MC-3550IPS

35

〜 50

3
6.3

less than 0.71

8.3
11.4

less than 0.75

13
18.3

Weight

105g
0.002

115g
4

140g
155g
305g

0.005

325g
465g

※ Three-point micrometer is precision instrument. For best performance, temperature should be around 20℃ with sufficient time to acclimate.
Extreme temperatures may affect accuracy and performance of measurement instruments.
※ Reference ring gauge is not included. Separate purchase is required.

221

念 校

04_201-226マイクロメータ.indd 221

CMYK

2019/03/18 16:28

19/03/18 16:28:02

MC-2025IPS
MC-2535IPS

2.8

95g

〜 20

151418

MICRO
METER
STAND

4

67

〜 8

151419

D

MICRO
METER

MICROMETER

DIGITAL THREE-POINT INTERNAL MICROMETERPDIProvided with an ring gauge for zero-point setting.
Ready to use immediately.

Reference standard
for calibration▶
Ring Gauge

Rust prevention▶
Zerust Film for Steel

P68
〜72

P380

● DIMENSIONS%RG\

MCD3385-1216HT

*$8*(352%(/&'',63/$<$77$&+0(17(;7(16,2152')25'((3(5+2/(

MCD3385-0608HT〜MCD3385-4050HTC

3 - GAUGE PROBE%RG\

MCD3385-7588HT

*$8*(352%(

LCD DISPLAY

A

B

MCD3385-5063HT〜MCD3385-87100HT/&'',63/$<$77$&+0(17(;7(16,2152')25'((3(5+2/(

Units: mm
Model No.

A

B

less than 2

85

C

D

MCD3385-0608HT
MCD3385-0810HT

100

デジタル三点マイクロメータ（50mm以上）
MCD3385-1012HT

MCD3385-1216HT

T_327

MCD3385-1620HT
MCD3385-2025HT

105

MCD3385-2530HT
MICRO
METER

19/03/18 16:28:05

MICRO
METER
STAND

USE
●Measurment of internal diameter
MATERIAL●Probe：Carbide tipped
FEATURES●Quick digital display of the measured value ●With ratchet stop
●Includes a setting ring ●Auto display off function（after approx. 5 min.）POWER●One SR44（Silver oxide）battery included for testing
Measuring Range（mm）

MCD3385-3040HT
MCD3385-4050HT

100less than 1

43
150

110

MCD3385-5063HT
MCD3385-6275HT

127

MCD3385-7588HT
MCD3385-87100HT※ Quantization Error（± 1 digit）not included
Resolution（mm）

Maximum Error ※（μm）

Order No.

Model No.

152221MCD3385-0608HT

6〜

8

180g

152222

MCD3385-0810HT

8 〜 10180g

Weight152223

MCD3385-1012HT

10 〜 12180g

152224

MCD3385-1216HT

12 〜 16230g

152225

MCD3385-1620HT

16 〜 20152226

MCD3385-2025HT

20 〜 25

4

230g
300g

0.001

152227

MCD3385-2530HT

25 〜 30

152228MCD3385-3040HT

30 〜 40320g

152229

MCD3385-4050HT

40 〜 50510g

152230MCD3385-5063HT

50 〜 63

710g

420g

5

152231MCD3385-6275HT

62 〜 75

152232

MCD3385-7588HT

75 〜 88720g
990g

152233

MCD3385-87100HT

87 〜 1001010g＊ Three-point micrometer is precision instrument. For best performance, temperature should be around 20℃ with sufficient time to acclimate.
Extreme temperatures may affect accuracy and performance of measurement instruments.

222

念 校

04_201-226マイクロメータ.indd 222

CMYK

2019/03/18 16:28

PDI

Provided with a reference ring gauge for zero-point setting.
Ready to use immediately.

Reference standard
for calibration▶
Ring Gauge

For holding micrometers▶
Micrometer Stand

P68
〜72

Rust prevention▶
Zerust Film for Steel

P226

P380

MICROMETER

DIGITAL INSIDE MICROMETER● DIMENSIONSMCD334-50I※ Measuring HeadMCD334-50I 〜 100I

USE
●For inside diameter measurement
MATERIAL●Frame: Carbon steel●Spindle: High carbon chrome steel
FEATURES●Quick digital display of the measured value●Provided with a reference ring gauge●With ratchet stop
●Auto display off function（after approx. 5 min.）●Carbide measuring faces
POWER●One SR44（Silver oxide）battery included for testing

MCD334-30I
Units: mm
Model No.

A

B

C

D

28

5

8

18

MCD334-30I
MCD334-50I
MCD334-75I
MCD334-100I※ Quantization Error（± 1 digit）not included

Order No.

Model No.

Measuring Range（mm）

151235

MCD334-30I

5 〜 30151236

MCD334-50I

25 〜 50

151237

MCD334-75I

50 〜 75

151238MCD334-100I

75 〜 100

Maximum Error ※（μm）

Resolution（mm）

Maximum Error（mm）

Weight

φ 5

330g

φ 25

330g

φ 50

300g

φ 75

320g

6
0.001
8

CALIPER TYPE INSIDE MICROMETERPDIProvided with a reference ring gauge for zero-point setting.
Ready to use immediately.

Reference standard
for calibration▶
Ring Gauge

For holding micrometers▶
Micrometer Stand

P68
〜72

Rust prevention▶
Zerust Film for Steel

P226

P380

● DIMENSIONSMC304-50I

USE●For inside diameter measurement
MATERIAL●Spindle: High carbon chrome steel
FEATURES●Provided with a reference ring gauge●With ratchet stop
●Carbide measuring faces

MICRO
METER
STAND

MC304-30I※ Measuring Head
MC304-50I 〜 100I
Model No.

MICRO
METER

Units: mm
A

B

C

D

28

5

8

18

MC304-30I
MC304-50I
MC304-75I
MC304-100I

Model No.

151230MC304-30I

5 〜 30151231MC304-50I

25 〜 50

151232

MC304-75I

50 〜 75

151233

MC304-100I

75 〜 100

Resolution
（mm）

0.01

Maximum Error（mm）

Weight

5

φ 5

140g

6

φ 25

150g

7

φ 50

160g

8

φ 75

190g

Maximum Error
（μm）

223

念 校

04_201-226マイクロメータ.indd 223

CMYK

2019/03/18 16:28

19/03/18 16:28:08

Measuring Range（mm）

Order No.

MICROMETER

DIGITAL DEPTH MICROMETERPDIFor high precision measurement of depth

Reference standard
for calibration▶
Block Gauge

Rust prevention▶
Zerust Film for Steel

P42
〜49

MCD233-25FA

USE●For depth measurement
MATERIAL●Base：Stainless●Probe：High carbon chrome steel
FEATURES●Quick digital display of the measured value●With ratchet stop
●Auto display off function（after approx. 5 min.）POWER●One SR44（Silver oxide）battery included for testingP380● DIMENSIONSUnits: mm
Model No.

A

MCD233-25FA

63

MCD233-25FB

101.5

B

C

D

E

9

4.5

18

17

※ Quantization Error（± 1 digit）not included
Order No.

Model No.

152181MCD233-25FA

152182

MCD233-25FB

Measuring Range（mm）

Resolution（mm）

Maximum Error ※（μm）

0 〜 250.001

6Weight

310g
380g

DEPTH MICROMETER

PDI
Reference standard
for calibration▶

For high precision measurement of depth

Block Gauge

Rust prevention▶
Zerust Film for Steel

P42
〜49

MICRO
METER

P380

MICRO
METER
STAND

● DIMENSIONSMC202-25F

19/03/18 16:28:12*$8*(
352%(

USE●For depth measurement
MATERIAL●Base：Stainless●Probe：High carbon chrome steel
FEATURES●With ratchet stop
Order No.

Model No.

151220MC202-25F

151221

MC202-25FS

Units: mm
Model No.

A

MC202-25F

63

MC202-25FS 101.5
Measuring Range（mm）

Resolution（mm）

Maximum Error（μm）

0 〜 25

0.01

4

B

C

D

E

9

4.5

18

17

Weight

210g
265g

224

念 校

04_201-226マイクロメータ.indd 224

CMYK

2019/03/18 16:28

PDI

For high precision measurement of depth.
Wide measuring range by interchangeable rods.

Reference standard
for calibration▶
Block Gauge

Rust prevention▶
Zerust Film for Steel

P42
〜49

P380

MICROMETER

INTERCHANGEABLE ROD TYPE DEPTH MICROMETER● DIMENSIONSMC202-100FS

USE●For depth measurement
MATERIAL●Base：Stainless●Probe：High carbon chrome steel
FEATURES●Able to provide a wide measuring range●With ratchet stop
●Interchangeable rod M
 C202-50F, MC202-50FS：2 pcs（25,50mm）
MC202-100F, MC202-100FS：4 pcs（25,50,75,100mm）
Order No.

Measuring Range（mm）

Model No.

152211

MC202-50F

152212

MC202-50FS

152213

MC202-100F

152214

MC202-100FS

Units: mm
Model No.

A

MC202-50F

63

MC202-50FS

101.5

MC202-100F

63

MC202-100FS

101.5

Resolution（mm）

0 〜 500.01

5

SETTING STANDARD

D

E

9

4.5

18

17

Weight

185g
265g
185g
265g

PDI

For origin setting of outside micrometers

A010-25

C

Maximum Error（μm）

4

0 〜 100

B

Rust prevention▶
Zerust Film for Steel

P380

A070-175

A020-50

A080-200

A030-75● DIMENSIONS

A010-25 〜 A060-150

A070-175, A080-200

USE●For accurate origin setting of outside micrometers（for size 25〜50 and larger size）
at the start or end of the measuring range
MATERIAL●Body: High Carbon chrome steel ●Grip: Plastic
Order No.

Model No.

Micrometer size

151491

A010-25

25 〜 50

151492

A020-50

50 〜 75

Accuracy（μm）±2Units: mm
Model No.

L

A010-25

25

A020-5050

A030-7575

A040-100

100

A050-125125

A060-150

150

A070-175

175

A080-200

200

Dimension（mm）

Weight

MICRO
METER
MICRO
METER
STAND

10g
φ7

15g

A030-75

75 〜 100

± 2.5

25g

A040-100

100 〜 125±345g151495

A050-125

125 〜 150

± 3.5

151496

A060-150

150 〜 175

±4

50g
φ8

65g

151497

A070-175

175 〜 200

± 4.5

75g

151498A080-200

200 〜 225

±5

85g

225

念 校

04_201-226マイクロメータ.indd 225

CMYK

2019/03/18 16:28

19/03/18 16:28:17

151493151494

Convenient measurement by holding objects in hands● DIMENSIONS

MS-RG

Units: mm

MS-SG
MS-RG

USE
●Stand for outside micrometers
MATERIAL●Body: Cast iron
FEATURES●Stable stand with moderate weight and low center of gravity●Rubber no-slip feet
●Clamp angle adjustable
SPECIFICATIONS●Clamp range：Outside micrometers up to 100mm

40°

°

75

73

6～14

21
115

Order No.

Model No.

Weight011766

MS-RG

1.3kg

011767

MS-SG

795g

44

MICROMETERMICROMETER STAND

MS-SG

COMBINATION STAND
Stand for micrometers as well as vernier calipers
MICRO
METER
MICRO
METER
STAND

● DIMENSIONSUnits: mm
45゜

22

85

35

3.5

0～18

19/03/05 10:38:18

160

USE
●Outside micrometer up to 100mm, vernier caliper up to 200mm
MATERIAL●Body: Cast iron
FEATURES●Stable stand with moderate weight and low center of gravity●Rubber no-slip feet
●Clamp angle adjustable 45°～90°

Clamp with grooves
to hold calipers
securely

Adjustable angle

Rubber no-slip feet

Order No.

Model No.

Weight011768

NMS-1G

1.5kg

226

三 校

04_201-226マイクロメータ.indd 226

CMYK

2019/03/05 10:38