Fall 2022 New ProductHighestaccuracy

in series history

3µm
Maximum error**DET-0800SW

DET-2000SW

DET-0800SW

Equipped with a new mechanismthat eliminates gears

DIGITAL S-LINE TEST INDICATOR
DET-2000SW／DET-0800SW

ycar
accutserhoigt
Hacidh＆nt
Iloaotitgeiul
DBepniyno
Buillotng Lever Tit
acinu
The aited
mmawotscelssbletaedrinaWsCto P■SPECIFICATIONS -Digital S-line Test Indicator

ResolutionDET-0800SW
DET-2000SW
2.0mm
0.8mm
36.5mm
12.5mm0.001mm

Minimum
reading display

0.01mm / 0.001mm (switchable)

Model No.
Measuring range
Measuring tip length

Maximum error
Hysteressis error

8µm
3µm

3µm
2µm
1µm

Repeatability

Digital DataManagement
Wirelesscommunication

Measuring force
Switching measurement
direction

software formeasuring instrument communication
SK-LOG integrated

Multiple measuring instruments management
Pass/ Fail judgement
Excel output

Less than 0.2N

Possible

Immersion-proof/ Dust-tight

IP54

Data communication

Built-in Bluetooth/ Optional data transfer cable (DTC-2)

Frequency band

2.4GHz（2.402 - 2.480GHz）
Class 3: 1mW（0dBm）
15 m or more（line of sight）

Maximum output
Communication
distance

Supports

Less than 0.1N

*1-5m in actual use environment

Mounting
shaft diameter

Attached stem：φ8mm

Operating temperature

5〜40°
CCR2032（lithium battery）×1（included for testing）
70g

PowerWeight
Other functions●Display rotates approx. 270°
●Simultaneous digital and analog display●MIN, MAX, DELTA (maximum displacement)
measurement●Measuring direction (+/-) switching●Preset●Auto display off (approx. 10 min.)*Our Bluetooth system is certified in Japan and FCC (USA), CE (EU), and KCC (Korea).
Please contact us for use in other countries.

MAGNETIC BASE WMB SERIES●Mechanical lock knob locks all joints with a single operation.
●Removable magnet with ON/OFF lever.

WMB-100G
WMB-80G
WMB-50G

Product name

Magnetic Base Lock Type

Model No.

WMB-50G

WMB-80G

WMB-100G

Dimension（W×D×H）

68×58×253mm

73×78×340mm

73×92×405mm

Base Dimension（W×D×H）

40×40×40mm

50×60×55mm

50×75×55mm

Magnetic adsorption

500N

800N
φ6mm ／ φ8mm ／ With dovetail grooveWeight710g

1,380g1,830gOrder No.

012896012897012898

Measuring instrument mounting hole

1,000N

2mm

Fine feed width

A single screw completes the positioning ofdial gauges and other devices to be installed!
Equipped with a fine adjustment function,
it is expected to shorten the measurement time!

This isfrom
diﬀerent
the conventional lock type! !
Quick mobility!

*It is possible to mount ameasurment instrument with
φ6mm・φ8mm stem/ dovetail plate.

Sturdy shaft!
Fine feed!

FEATURES

Install the dial gauge.

Powerfull locking
with light force!

Strong grippingpower!
Easy to ON/OFF!

For holding
cylindrical workpieces!

Strongmagnetic force!

*Use the optional cylindrical clamping jig CC-WMB.

*Size and Weight are approximate value.
*Appearance and speciﬁcations are subject tochange without notice for productimprovement.
*Colors may vary slightly between printed andactual product.

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

No.22102000PA■DistributorsAnodized
aluminum body!