PRODUCT INFORMATION

EM7000 - Communication Board for Gas Sensor
Evaluation Modules

Features:* Communication interface for transferring output signals from Figaro sensor evaluation
modules to an external PC* PC application software for importing and saving gas measurement data on a PC
* Real-time graphic display of gas concentration or output voltage of the gas sensors on aPC monitor* Easy control of gas measurement operations through the control panel on a PC monitor

Description:

The EM7000 communication board is designed to transfer output from EMseries sensor evaluation modules to an external computer for facilitating
evaluation of the characteristics of various Figaro gas sensors. Equippedwith a USB connector to transfer output data from the EM series evaluation
modules to a PC via serial communication, an output terminal for gas
concentration data in PWM format from the CDM series CO2 sensor, and aterminal for voltage output converted from PWM output, this communication
board allows for a quick and easy start to evaluate the characteristics ofFigaro gas sensors.
Once the PC application software is installed on an external PC, it allows
for the transfer of measurement data in serial format to the PC and thegraphic display of output data on the PC monitor. Measurement data can beimported and saved in CSV format on the PC using the application softwareand then utilized for data processing and analysis after gas measurements using other software, such as Microsoft
Excel (Note 3).
The communication board includes a USB memory stick containing PC application software and a Mini-B USB
cable. Please note that Figaro gas sensors or EM series evaluation modules are not included, but can bepurchased separately depending on evaluation needs.

PC Application Software:
Operating System

Japanese and English
language version Windows®
10 (Note 3)

Graphic displayof measured data

0 - 5,000ppm
for CO2 sensor modules
0 - 5Vfor other gas sensors andevaluation modules(according to thespecifications of each EM
series modules)

Data file format

CSV

Data saving
interval
Data filecapacity

2, 30, 60 sec.,
6, 10, 30, or 60 min.
Measurement data for up toapprox. 24 hours per eachfile* Downloading the Measurement Data Management Software
Download a set of the PC Application software for EM7000 from the following website of Figaro
Engineering Inc. onto a PC.

https://www.figaro.co.jp/en/product/entry/em7000.html

PRODUCT INFORMATION
Specifications:
Product Name

Communication Board for Gas Sensor Evaluation Modules

Model No.

EM7000Compatible evaluation modules
and sensors

EM series evaluation modules, CDM7162 (Note 2)

Power Supply

via USB Connector or 5.0V DC

Serial Output

0~65,535ppm CO2 sensors
Output varies according to specifications ofEM series evaluation modules other than CO2 sensor. (Note 2)

PWM Output

0 – 100% duty cycle for 0 - 5000ppm for CO2 sensors only

Voltage Output

0 - 5VDC for 0 - 5000ppm for CO2 sensor module
Output varies according to specifications of EM series evaluation modules.

Alarm Output

CMOS output for CO2 sensors only
(High: @CO2 conc. > 1000ppm Low: @CO2 conc. < 900ppm) (Note 2)

Operating Conditions

0~50˚C/0~95%RH (no condensation) excl. CDM7162 (Note 2)

Dimensions

approx. 60 x 68 mm
Height includes spacers, excluding protruding parts.

Standard Accessories

1 pce. mini-B USB cable

Dimensions:
1.6

60EM7000

ALARM

1CN1

PWM

56CN214CN43 4SW4323mini-B USB conector

0VJ1OFFSW1*Gas sensors, CO2 sensors, and EM series evaluation
modules are available for purchase from Figaro Engineering.

D2CN3

D1UART
I2C

USB5V INON

3SW2

1

D3

SW31

521

ON

1CN6

FIGARO

CN5

3.5
68

104-φ3.2

3.5

Example of the communication board fittedwith an evaluation board and a CO2 sensor:DC jack

11Unit : mmNote 1) For the evaluation of the CO2 sensor module CDM7162, please solder terminal pins available
in the marketplace to the CDM7162 to attach it to the communication board.
Note 2) Please refer to the Product Information of each gas sensors and EM series evaluation modules
for specification details.
Note 3) Windows and Excel are registered trademarks of Microsoft Corporation in the United States.

FIGARO ENGINEERING INC.
1-5-11 Senba-nishi
Mino, Osaka 562-8505 JAPAN
Phone: (81)-727-28-2045
URL：www.figaro.co.jp/en/

REV: 03/24