Outward dimensions（Unit: mm）

Outward dimensions（Unit: mm）
CB-X013-□□（Waterproof cable）

HM-0013-M (Long period vibration monitoring unit）

TM-HMST (Wall mounting bracket) *Optional accessory

35.8

Installation level

Tilt angle 40 °

φ12.3

1570

12LAN
connector

4210

6

USB port

8

Left side

6Front100

10

Relay output
connector

Side lid

1214

Top

16100

6190

Pullingdirection

20

BackSensor lid

1251

CH1-NO

3

CH1-C
CH2-NO

5

CH2-C
CH3-NO

7

CH3-C
CH4-NO

9

CH4-C
CH5-NO

11

CH5-C
CH6-NO

13

CH6-C
CH7-NO

15

CH7-CCH8-NO

TM-0013-HM

37.5

Long period vibration monitoring unit (HM-0013 series) + Smart display unit (TM-0013-HM)

HM-PB01 (Outdoor protection box for floor mounting)*Optional accessory

HM-PB01ST (Outdoor protection box for wall mounting)Multi-point earthquake
monitoring system*Optional accessory

Earthquake Watcher

Base plate

220
220

109

CH8-C

109

Bottom side（No sensor lid）

RP-0013 (Power relay BOX) *Optional accessory

4737.5227

HM
connector
Right side

HM-0013232

Optional alarm output cable

Bottom64-C5（200）

Relay output
connector
2

Externalmonitor port

291.5-391.5(Stepless）

Terminal for M3 screw(round type)Internal:
Relay circuit

R34-φ5Pedestal 350°

284R7.5
100

R3246-346(Stepless）*The terminal unit HM-0013-E has the terminals
on the lower right side of this figure sealed.

65

4-R5

t=2.3

26TM-0013-HM (Smart display unit）

6R3

R7.5

135 110
Tilt angle -5°

6R32575 91

90

12.50

160
100

612.775

Cable outer
diameter φ5.1

120

91

12.5

40.50

CR-27BK (Stand for HM-0013) *Optional accessory

TB-0013 (Cable relay box)

220

Base plate*Optional accessory

3032.50220OUT

888010895

IN4368【Side】【Top】UC-0013 (USB conversion box for HM-0013)) *Optional accessory
7.4

Head Office
95tel
webPUSH+81 6 6471 3155
https://www.imv.co.jp/e*The specifications and design are subject to change without notice.

95

3DSS Business Division
MES Business Division
IMV Advanced Technology Research Laboratory
Sales Division
Quality Assurance Department
Personnel General Affairs Department

2528.8CN/14640E
Osaka site

Widely monitoring from long period earthquake to strong earthquake

JQA-1573Jan. 2021

As soon as the number of observation points and cable length are determined,
By combining the long-period vibration monitoring unit and the smart display unit, it is possible toconstruct a multi-point earthquake monitoring and observation system.

Smart display unit

Long period vibration
monitoring unit

actual measurement starts!
Function

Easy to operate with easy-to-read icons

Specifications
Warning flash screen that allows simultaneousconfirmation of multiple points

TM-0013-HM● Accurate detection of long cycle vibration+Item
NoteHigh precision mode High frequency mode
3axis (X,Y,Z）
Measurement axis
±2000 Gal±6000 GalEach axis
Measurement range (full scale)up to100Hz
Sensor only 1000 Hz
Frequency range
Acceleration accuracy ±0.5 %FS（±0.1%FS） ±5.0 %FS（±1.0 %FS） Standard (special)
Sensor noise density (rms)
A/D resolution
Power supplyStructure
Communication
Time accuracy
Ambient temperaturePower consumption
Size
WeightFunction

HM-0013-M（Middle unit）
HM-0013-E （End unit）

Long period vibration monitoring system (Common to HM-0013-M, HM-0013-E)● Up to 4 HM-0013 can be connected● Measurement of fine movement of 0.1 gal is possible● 7 inch touch panel installed● Data save memory for power failure installed● Record waveform by time reservation● Protection grade IP67/Shock resistance 1000G● Supports email notification via Ethernet

XY: 0.0003 g / √Hz lebel
Z: 0.0005 g / √Hz lebel

24 Bit
DC12 V±20 ％
IP67
RS485 (Modbus）
5.0 ppm-10 to＋60 ℃
30 mA
90（H) × 70（W) × 40（D) mm
Approx. 400 g
Temperature correction
Temperature output

When an earthquake occurs, the seismic intensity,
maximum acceleration, and time of occurrence detectedby each HM-0013 are displayed.

Sensor self check

Connection
example・Main unit setting file・Waveform data

LAN (Ethernet) cable

5-20 m地震検知
ノード1

・Time correction (NTP server)
・FTP file transfer

発生日時

アドレス102.20.2016

震度4653.3

ノード3

アドレス3

発生日時

02.20.2016

震度
最大加速度

4638.2

ノード2
11：00：36

7

最大加速度

Gal

11：00：36

7

Gal

現在日時 02.20.2016 12：00：00

発生日時

アドレス202.20.2016

震度4624.3

ノード4

アドレス4

発生日時

02.20.2016

震度
最大加速度

4645.1

11：00：36

7

最大加速度Record up to 5000 records

4 connections: Maximum cable length 80 m*

5-20 m

* Depending on memory card capacity

Status monitoring with reserved measurement

5-20 mGal

Convertible in option
Accurate with 13 seconds per month
At DC12 V
Excluding connectors etc.

Near temperature of sensor module
Sensor condition confirmation

Smart display unit (TM-0013-HM)
Specifications

Clock

Accuracy: 20 ppm (daily difference 2 seconds) or lessTime synchronization of the main clock is possible with NTP, GPS, etc.

Installation method

VESA standard 100 mm x 100 mm Installation hole
Wall Installation bracket (optional)

Ambient temperature
Ambient humidity
Power supply

0 to 40 ℃
20 to 45 ％RH (Non-condensing）
AC100-240 V ±10 ％50 VA or less（When 4 pcs of HM-0013 are connected）

Size
Weight210(H) x 284（W) x 65（D) mm (excluding connectors etc.)
Approx. 2 kg

・Time correction (GPS)

7HM-0013-M

HM-0013-M

HM-0013-M

HM-0013-E(Middle unit）(Middle unit）(Middle unit）(End unit）*If the power relay BOX (RP-0013) isconnected between the monitoring units,
the distance after that can be used up to 80 m.

Example of installation

RP-0013

Building
Upper floor

Item
Size
Display method
Drive system
Back light
Touch panel method

Up to 5000 recorded data * can be recorded.
*Depends on the capacity of the memory card.
The history can be checked in the list, and the waveform
screen can be checked by clicking.

Up to 4 waveform recordings can be reserved dependingon the settings, and the condition of the building can bemonitored on a regular basis.

Can be analyzed from the waveform display screen

Easy data collection with memory card

Item
Connector shape
The number of pins
Bit rate setting
Connector internal
pin placement

Item
Connector shape
The number of pins
Contact specifications

Connector internal
pin placement

Power relay box(if necessary)

ACadaptor

RP-0013

AC adaptor

RP-0013First floor

HM-0013-M(Middle unit）

HM-0013-M(Middle unit）

Specifications
One-touch lock connector
16 pins
The number of contacts 8（AQV252GA made by Panasonic )Contact rating 60 V-2.5 A (AC/DC both use, peak value)
Contact function Linked to vibration detection of each HM-0013
Contact operation Open during standby/Closed when vibration is detected

Pin No.
1/2
3/4
5/6
7/8
9/10
11/12
13/14
15/16

Signal name
CH1-NO/CH1-C
CH2-NO/CH2-C
CH3-NO/CH3-C
CH4-NO/CH4-C
CH5-NO/CH5-C
CH6-NO/CH6-C
CH7-NO/CH7-C
CH8-NO/CH8-C

LAN (Ethernet) Possible with a commercially available cable with the following specifications
地震検知
ノード1
発生日時

アドレス1

アドレス302.20.2016 11：00：36

震度4638.2

アドレス1

ノード202.20.2016 11：00：36

震度7

最大加速度4653.3

ノード3

アドレス3

発生日時

02.20.2016 11：00：36

震度
最大加速度

4638.2

Gal

7

Gal

12：00：00

発生日時

アドレス202.20.2016 11：00：36

震度7

最大加速度4624.3

ノード4

アドレス4

発生日時

02.20.2016 11：00：36

震度
最大加速度

4645.1

Gal

7

Gal

12：00：00

発生日時

アドレス202.20.2016 11：00：36

震度7

最大加速度4624.3

ノード4

アドレス4

発生日時

02.20.2016 11：00：36

震度
最大加速度

4645.1

Gal7Gal戻る

B floor

地震検知
ノード1
発生日時

74653.3

ノード3

最大加速度

現在日時 02.20.2016

ノード202.20.2016 11：00：36

震度最大加速度

発生日時

現在日時 02.20.2016

HM-0013-M(Middle unit）

Specifications
One-touch lock connector
6 pins781250 bps/115200 bps [default]/57600 bps/38400 bps
Pin No.
Signal name （HM-0013-M/E)
1/2
Power supply +/Power supply 3/4
D-OUT/D-IN
5/6
RS-485+/RS-485-

Contact output [Relay output] System configuration diagram

HM-0013-M(Middle unit）

HM-0013-M(Middle unit）

Power relay box(if necessary)

Gal

7Gal

戻る

TM-0013-HM

SpecificationsDC4.5 V to 9 V
15 V/250 mA-10 to 60°C80（H) x 47（W) x 32.5（D) mm
150 g*With AC adapter (2 flat pins), power supply voltage 100-240 VAC, ripple 10%

BU-353S4 (USB-GPS reciever) *Optional accessories
Item
Size
Connector shape
Cable length
Ambient temperature
OtherSpecifications
Main unit: φ Approx. 60 ｘ 20 （D) mm
USB connector
152 mm-40 to 80℃, 5 to 95％ (Non-condensing）
With bottom magnet

TB-0013 (Cable relay box) *Optional accessories
Item
Size
Storage cable diameter
Internal terminal block
Material

Specifications
88 (H) x 68 （W) x 43 （D) mm (excluding connectors etc.)
φ12 to 14.4 mm
M3 screw terminal 6P, M4 screw ground terminal
Lid/body: Aluminum alloy die-cast

TM-0013-HM

HM-0013-M(Middle unit）

Acceleration waveform confirmation and FFT analysis ofrecorded data are possible. It is possible to always grasp
the state of the building at normal times, of course, after
the earthquake.

The data is recorded on the upper memory card and canbe taken out and replaced smoothly. Even if the display isdamaged, the data can be retrieved if the record remains
on the card.

ItemConnector shapeStandardUse

ItemSize
Mounting hole position
Withstand load
Material*Optional accessories

Specifications
291.5 - 391.5 (H) x 232 （W) x 227 （D) mm
VESA standard (100 x 100 mm 4 places)
3 to 6 kg
PS

UC-0013 (USB conversion box for HM-0013)

HM-0013 Connection [HM] System Composition Diagram

HM-0013-E(End unit）

Automaticwarehouse

Specifications
7.0-inch wide, 800 (H) x 480 (V) dots
TN 262, 144 color
TFT active matrix
High-brightness white LED (OFF time can be changed by setting)Resistive film analog tabletInterface

HM-0013-E(End unit）

Middle floor

Item
DC IN Power supply voltage
Power supply
Ambient temperature
Size
Weight

CR-27BK (Stand for TM-0013)

戻る□□・Linked to vibration detection of each sensor

RP-0013 (Power relay box) *Optional accessories

Gal

Dedicated waterproof cable CB-X013: diameter5.1 mm
Connector diameter at both ends: 12.3 mm

Alarm output cable CP-TMRO-5 (optional)□□)Type
Specifications
Conductor
Tin-plated annealed copper stranded wire
CB-X013-05
Special elastomer
CB-X013-10 Cable Insulator
partSheath material Oil resistance PVC
CB-X013-20Ambient temperature -10 to 80° C
CB-X013-40
Zinc alloy nickel plating
CB-X013-80 Connector Exterior
part
Ambient temperature -25 to 85°C

11：00：36

Display part

USB GPS receiver BU-353S4 (optional)

Name
Waterproof cable 5 m
Waterproof cable 10 m
Waterproof cable 20 m
Waterproof cable 40 m
Waterproof cable 80 m

ΔΣ type* Be sure to use the terminal unit HM-0013-E at the end of the monitoring unit connection.
Otherwise, A communication error may occur.

ItemMemory card

0.1 Hz to 100 Hz (Less than 0.00004 g /√Hz
between 1 Hz to 100 Hz）

Daisy chain connection

By adopting a color touch panel and menu display, it ispossible to easily display and set the system status andwaveform data.

System Conposition

XY: 0.0001 g / √Hz lebel
Z: 0.0002 g / √Hz lebel

Dedicated waterproof cable (CB-X013-Specifications
ＲＪ４５ connector
１０ＢＡＳＥ-Ｔ/100ＢＡＳＥ-ＴＸ

FTP client function: Automatic file transfer
NTP client function: Time synchronization of the main clock

Item
PC connection interface
DC IN Power supply voltage
Power supplyContact
Recommended contact plug
Top button
Sensor switch
Ambient temperature
SizeWeight*Optional accessories

Specifications
microUSB connector
DC 4.5 V to 9.0 V
12 V/250 ｍA
12 V/80 ｍA
1ch0395000002 mad by Molex
External trigger pin interlocking
Power ON/OFF of connection HM-10 to 60 ℃
95 (H) x 95 （W) x 28.8 （D) mm
Approx. 230 g

Note
USB2.0
Dedicated AC adaptor AC100 V/240 V
When connecting the AC adapter
When driven by USB bus power
60 V/2.5 A
3.50 mm pitch

excluding connectors etc.

* When interlocking measurement is performed using the nearest node as a trigger,
it is necessary to modify UC-0013 to loopback mode. Please check the measurement method
before purchasing and order the required mode.