LCR Elite1

an efficient, convenient & accurate LCR metre
LCR Elite1 is a portable LCR metre with unique mechanical and electronic design. It integrates apair of tweezers like probes and a digital LCR metre into one compact, lightweight, battery
powered device.

Simple Navigation Button:
Automatic or manual selection for component
type, test frequencies and circuit modes. A single
button allows access to four different modes.

Crisp OLED Display:
The primary display shows component
type and value. The secondary display
shows Rs or Rp. Testing parameters andbattery indicator are also displayed.

Ergonomic Design:Standard USB Charging:
The built-in rechargeable lithium-ion
polymer battery can be conveniently
charged using your computer or a
USB power adapter.

The portable design allows user
one handed operation toimprove productivity.
Precise Gold Plated Tips:
Pick up and measure SMD resistors,capacitors and inductors easily,reliably and accurately.

Made in Canada

Sophisticated Design for SMD Component Measurement:
LCR Elite1 is designed to measure inductance, capacitance and resistance with high accuracy. It provides a simple and efficientsolution for measuring and identifying SMD components as well as troubleshooting electronic circuits.
The gold-plated precise tips are designed to contact the SMD components easily and reliably with the size down to 0201.
The 4-wire shielded probe makes parasitic parametres small and very predictable. It improves measurement accuracy andsignificantly reduces the probability of measurement errors related to setup.
The compact design allows user one hand operation and makes a convenient way to take measurements and read results.

Automated Component Identification
LCR Elite1 simplifies measurements by using the automatic component identification function. It automatically identifies L, C, or
R and selects proper testing frequency and circuit mode (parallel or series). Alternatively, user can use manual mode to set thedesired parametres for measurement.
Detailed component analysis is provided on the OLED display. The primary display shows component type and value.
The secondary display shows Rs (series equivalent resistance) or Rp (parallel equivalent resistance). Testing parameters andbattery indicator are also displayed.

Easy Operation
LCR Elite1 provides shortcut to go to the default mode quickly. As long as the navigation button is pressed down for
approximately 2 seconds, the device goes to the default mode no matter which mode it is currently in. The device can be turned
off as long as the navigation button is pressed down longer than 5 seconds.

Long Battery LifeLCR Elite1 is powered by an internal, lithium-ion polymer rechargeable battery. It can be charged by a computer or a USB power
adapter (available separately).
The power consumption is optimized to make the battery last a day for typical measurement. The standby time can be longer
than a whole year. It makes the device an ideal choice for broad range of applications and missions, including R&D labs,
production lines, service and repair, etc.
Testing Signal

Product Characteristics

Test frequency:

100 Hz, 1 kHz, 10 kHz

Size:

151 x 19 x 14.5mm

Test signal level:

0.45Vrms

Weight:30 grams

Source impedance:

100 Ω ± 1%

Operating temperature:-10°C to 50°C

Battery Type:LiPO rechargeable,
3.7V 150mAH

Measurement Range
Resistance R:

25 mΩ to 10 MΩ

Battery Life:All day in typical measurement

Inductance L:

1.0 uH to 1 H

Charging time:

2.5 hours typical

Capacitance C:

0.3 pF to 500 uF

Learn more at:www.lcrresearch.com

Resistance R:

0.5 %email us at: sales@lcrresearch.com

Inductance L:

1.0 %Capacitance C:

1.0 %Basic Accuracy ** For more detailed information, please refer to LCR Elite1 User Manual.

LCR Research Ltd.
660 Eglinton Ave East, Sunnybrook PO Box 50207,
Toronto, Ontario, Canada M4G 0B5© LCR Research Ltd. 2015. All rights reserved.