Digital Blood Pressure Monitor

Model UA-1030
Instruction Manual
Original

Manuel d’instructions
Traduction

Manual de Instrucciones
Traducción

Manuale di Istruzioni
Traduzione

使用手冊

翻譯

1WMPD4002225C

Contents
Dear Customers ..............................................................................................1
Preliminary Remarks........................................................................................1
Precautions .....................................................................................................2
Parts Identification ..........................................................................................3
Symbols ..........................................................................................................4
Operation Mode ..............................................................................................6
Using the Monitor............................................................................................7
Installing / Changing the Batteries ............................................................7
Connecting the Air Hose ...........................................................................7
Connecting the AC Adapter .......................................................................7
Adjusting the Built-in Clock ......................................................................8
Selecting the Correct Cuff Size ..................................................................8
Applying the Arm Cuff ..............................................................................9How to Take Accurate Measurements ........................................................9
Measurement ............................................................................................9
After Measurement....................................................................................9
Measurements...............................................................................................10
Normal Measurement ..............................................................................10
Measurement with the SET Pressure .......................................................11
Measurement with the Desired Systolic Pressure .....................................11
Notes for Accurate Measurement.............................................................12
TriCheckTM Measurement...............................................................................12
Recalling the Memory Data ............................................................................14
Adjusting the Speaker Volume.......................................................................15
Selecting the Language..................................................................................15
What is an Irregular Heartbeat .......................................................................16
Pressure Bar Indicator....................................................................................16
WHO Classification Indicator..........................................................................16
About Blood Pressure ....................................................................................17
What is Blood Pressure?...........................................................................17
What is Hypertension and How is it Controlled? .......................................17
Why Measure Blood Pressure at Home?....................................................17
WHO Blood Pressure Classification ..........................................................17
Blood Pressure Variations ........................................................................17
Troubleshooting............................................................................................18
Maintenance..................................................................................................19
Technical Data...............................................................................................19

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure monitor,
one of the most advanced monitors available today. Designed for ease of use
and accuracy, this device will facilitate your daily blood pressure regimen.
We recommend that you read through this manual carefully before using
the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity.
(0123: The reference number to the involved notified body)
The device is designed for use on adults only, not newborns or infants.
Environment for use: The device is for use in the home healthcare environment.
This device is designed to measure blood pressure and pulse rate ofpeople for diagnosis.
English 1

Precautions
Precision components are used in the construction of this device. Extremesin temperature, humidity, direct sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff or storing the hose tightly twisted for long
periods, as such treatment may shorten the life of the components.
The device and cuff are not water resistant. Prevent rain, sweat and waterfrom soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
Used equipment, parts and batteries are not treated as ordinary household
waste, and must be disposed of according to the applicable local regulations.
When the AC adapter is used, make sure that the AC adapter can bereadily removed from the electrical outlet when necessary.
When reusing the device, confirm that the device is clean.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the arm must be squeezed by the cuff hardenough to temporarily stop blood flow through the artery. This may cause
pain, numbness or a temporary red mark to the arm. This condition willappear especially when measurement is repeated successively.
Any pain, numbness, or red marks will disappear with time.

Contraindications
The following are precautions for proper use of the device.
Do not apply the cuff to an arm with another medical electrical equipment
attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult adoctor before using the device, to avoid medical problems.
Do not self-diagnose the measurement results and start treatment by yourself.
Always consult your doctor for evaluation of the results and treatment.
Do not apply the cuff on an arm with an unhealed wound.
Do not apply the cuff on an arm receiving an intravenous drip or bloodtransfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as ahigh-pressure oxygen chamber or an oxygen tent. It may cause a fire or explosion.

English 2

Parts Identification
DC Jack

Speaker

Display
ButtonButtonSET Button

Air Connector Plug
Arm Cuff

VOLUME
ButtonSTART Button
Air Hose

Air Socket

Battery Compartment

Battery Cover

Display
MEMORY

Average

1.5V Batteries(R6P, LR6 or AA)

Pressure Settings

I.H.B. Symbol(Irregular heartbeat symbol)
Cuff Fit Error Symbol

Systolic
Pressure

Movement Error Symbol
TriCheckTM Symbol

Diastolic
PressureHeart MarkWHO Classification Indicatorand Pressure Bar Indicator
Battery Indicator

Pulse Rate

AM / PM MarkEnglish 3

Clock Display

Symbols
Symbols that are printed on the device and the AC adapter
Symbols

Function / Meaning

Recommended Action

Standby and Turn the device on.
Battery installation guide
Direct current
SN

Serial number

2010

Date of manufacture
Type BF: Device, cuff and tubing aredesigned to provide special protection
against electrical shocks.
EC directive medical device label
WEEE label
Manufacturer
EU-representative
Refer to instruction manual/booklet
Class II device
Polarity of DC jack

Symbols that appear on the display
Symbols

Function / Meaning
Recommended Action
Appears while measurement is in progress. Measurement is in progress.
It blinks when the pulse is detected.
Remain as still as possible.
Irregular Heartbeat symbol (I.H.B.)
Appears when an irregular heartbeat
is detected.
It may light when a very slight vibration
like shivering or shaking is detected.
TriCheck mode
Takes three consecutive measurements
automatically and displays the averagevalues of the three measurements.
The reading may yield
an incorrect value. Take
Appears when a body or arm
another measurement.
movement is detected.
Remain still during
measurement.
English 4

Symbols

Function / Meaning

Recommended Action

Appears during measurement when
the cuff is applied loosely.

The reading may yield anincorrect value. Apply the
cuff correctly, and takeanother measurement.

Previous measurements stored in memory.
Average data
Full Battery

The battery power indicator during
measurement.

Low Battery

The battery power is low when it
blinks.

Replace all batteries with newones when the mark blinks.

Unstable blood pressure due tomovement during measurement.

Take another measurement.
Remain still during
measurement.

The systolic and diastolic values arewithin 10 mmHg of each other.
The pressure value did not increaseduring the inflation.
The cuff is not applied correctly.
PUL. DISPLAY ERROR

The pulse is not detected correctly.

Blood pressure monitor internal error

SYS.

Systolic blood pressure in mmHg

DIA.

Diastolic blood pressure in mmHg

PUL./min.

Apply the cuff correctly,
and take anothermeasurement.

Pulse per minuteAM

Data taken between 4:00 and 9:59

PM

Data taken between 18:00 and 1:59
Pressure settings
Indicates the pressure value
previously set by the user.
English 5

Remove the batteries andpress the START button,
and then install the batteries
again. If the error still
appears, contact the dealer.

Operation Mode
1. Normal Measurement
Press the START button. Blood pressure is measured and the data is stored
in memory. This device can store the last 90 measurements in memory.

2. Recalling the Data
Press theor button to recall the data inmemory. The average of all measurements is
displayed, as indicated in the figure at the right.
The device announces the memory data
values as they are displayed.
button is pressed, theThen, each time thememory data is displayed as follows.
Average of all AM (morning) measurements
taken between 4:00 and 9:59.
Average of all PM (evening) measurements
taken between 18:00 and 1:59.
Most recent data (No.n, in the example,
No.35)
Last data (No.1)
For details on recalling the data, refer to“Recalling the Memory Data”.

3. Deleting all Data Stored in Memory
Press both theand buttons. The
markand the battery indicator appear. Press and hold
both theand buttons until the illuminatedmark starts blinking to delete all data stored inmemory.

4. Measurement with the Desired Systolic Pressure
Refer to page 11 for measurement with the desired systolic pressure.
English 6

Using the Monitor
Installing / Changing the Batteries
1. Remove the battery cover.
2. Remove the used batteries andinsert new batteries into the battery
compartment as shown, taking
care that the polarities (+ and -) arecorrect.
Use only R6P, LR6 or batteries.
3. Attach the battery cover.

Step 1
Step 3

Step 1
Step 2

CAUTION
Insert the batteries as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(LOW BATTERY mark) blinks on the display and the device
announces that the battery needs to be replaced, replace all batteries withnew ones. Do not mix old and new batteries. It may shorten the battery life,
or cause the device to malfunction.
Replace the batteries two seconds or more after the device turns off.
If
(LOW BATTERY mark) appears even after the batteries are replaced, make a
blood pressure measurement. The device may then recognize the new batteries.
(LOW BATTERY mark) does not appear when the batteries are drained.
The battery life varies with the ambient temperature and may be shorter at
low temperatures.
Use the specified batteries only.
Remove the batteries if the device is not to be used for a long time.
The batteries may leak and cause a malfunction.
Air socket

Connecting the Air Hose
Insert the air connector plug into
the air socket firmly.

Air connector plug

Connecting the AC Adapter
Insert the AC adapter plug into the
DC jack.
Next, connect the AC adapter to anelectrical outlet.
Use the specified AC adapter.
(Refer to page 20.)

English 7

AC adapter plugDC jack

Using the Monitor
Adjusting the Built-in Clock

Press

Adjust the clock prior to use.
1. Press the SET button until the year starts
blinking.
or button.
2. Select the year using thePress the SET button to set the current year
and move to month/day selection. The datecan be set anywhere between the years 2010
and 2059.
or button.
3. Select the month using thePress the SET button to set the current
month and move to day selection.
or button.
4. Select the day using thePress the SET button to set the current day
and move to hour/minute selection.
or button.
5. Select the hour using thePress the SET button to set the current hourand move to minute selection.
or button.
6. Select the minute using the(Press the SET button to go to the pressure
setting mode. For details, refer to page 11.)
Press the START button to turn the device off.

Year

or
Month

Day

or

or
Hour

MinuteNote: After three minutes of non-operation, the device
will turn off automatically.
When the clock has not been
Turn off
set, the clock display indicatesdashes as shown to the right.
Pressing the START
button will turn theHolding down theorbuttondevice off anytime.
will change the value continuously.

Selecting the Correct Cuff Sizeoror

Pressuresetting mode

Using the correct cuff size is important for an accurate reading. If the cuff is not
the proper size, the reading may yield an incorrect blood pressure value.
The arm size is printed on each cuff.
The arm cuff is a consumable. If it becomes worn, purchase a new one.
Arm Size
Recommended Cuff Size
Catalog Number
31 cm to 45 cm
Large adult cuff
CUF-F-LA
17 cm to 32 cm
Adult cuff
CUF-G-A
Arm size: The circumference of the biceps.
English 8

Using the Monitor
Do not roll up shirtsleeve tightly.

Applying the Arm Cuff

1. Wrap the cuff around the upper
arm, about 1-2 cm above theinside of the elbow, as shown.
Place the cuff directly against theskin, as clothing may cause a faintpulse, and result in a measurement
error.
2. Constriction of the upper arm,
caused by tightly rolling up a
shirtsleeve, may prevent accurate
readings.

Cuff

1-2 cmAir Hose

Artery position markSecure the fabric fastener
firmly.

Symbols that are printed on the cuff
Symbols
Function/Meaning
Recommended Action
mark on the artery of the upper
Set thearm or in line with the ring finger on theArtery Position Markinside of the arm.
REF

Catalog Number
Lot Number

How to Take Accurate Measurements
For the most accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross yourlegs. Keep your feet on the floor and straighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same level as your heart.
Remain still and keep quiet during measurement.
Do not measure immediately after physical exercise or a bath. Rest fortwenty or thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

Measurement

During measurement, it is normal for the cuff to feel very tight. (Do not be
alarmed).

After Measurement

After measurement, press the START button to turn the device off. After one
minute of non-operation, the device will turn off automatically.
Remove the cuff and record your data.

Note: Allow at least three minutes between measurements on the same person.
English 9

Measurements
Before measurement, read “Notes for Accurate Measurement” on page 12.

Normal Measurement
1.

Place the cuff on the arm (preferably the left arm).
Sit quietly during measurement.

2.

Press the START button.
Press
All of the display segments are displayed.
Zero (0) is displayed blinking briefly.
The display changes, as indicated in the
All of the display
figure at the right, as the measurement segments displayed
begins. The cuff starts to inflate. It is normal
for the cuff to feel very tight. A pressure bar
indicator is displayed, on the left edge of thedisplay, during the inflation.
Zero display
Note: If you wish to stop inflation at any time,
Starts inflationpress the START button again.

3.

When inflation is complete, deflation starts
automatically and(heart mark) blinks,
indicating that the measurement is inprogress. Once the pulse is detected, themark blinks with each pulse beat.
Note: If an appropriate pressure is not obtained, thedevice starts to inflate again automatically.
To
avoid
re-inflation,
see“Measurement with the SET
Pressure” on the next page.

4.

When the measurement is complete, thesystolic and diastolic pressure readings
Measurement
and pulse rate are displayed. The device
in progress
announces the measurement results.
The cuff exhausts the remaining air anddeflates completely.
Systolic pressure
Diastolic pressure
Press the START button to turn the device off.
WHO classification
After one minute of non-operation,
the device will turn off automatically.
Pulse rate

5.

Note: Allow at least three minutes between
measurements on the same person.

English 10

At heart level
START

Pressurizing

Exhausts remaining
air automaticallyMeasurements
Model UA-1030T is designed to detect the pulse and to inflate the cuff toa systolic pressure level automatically.
If re-inflation occurs repeatedly, use the following methods.

Measurement with the SET Pressure
During the blood pressure measurement, re-inflation may occur.
A fixed pressure value can be set to avoid re-inflation.
1.

2.

At step 6 of the clock adjustment procedure(see page 8), press the SET button to go to
the pressure setting mode. The current
setting blinks.

orPress theorbutton to select a pressure
value about 30 mmHg or more above yourexpected systolic pressure from the following.
AUTO : Automatic pressurization (default value)
180 : Pressure value of 180 mmHg (fixed)
210 : Pressure value of 210 mmHg (fixed)
Turn off
240 : Pressure value of 240 mmHg (fixed)

3.

TriChecksetting mode

Press the SET button to go to the TriCheck setting mode.
Press the START button to turn the device off. After three minutes of
non-operation, the device will turn off automatically.
The next measurement will be performed with the new pressure value.

Measurement with the Desired Systolic Pressure
Use this method when re-inflation occurs repeatedly even if the pressure value
is set to 240 in the procedure above or when the results are not displayedeven if the pressure decreases to 20 mmHg or less.
1.

Place the cuff on the arm (preferably the left arm). At heart level2.

Press and hold the START button until a
number about 30 to 40 mmHg higher
than your expected systolic pressure
appears.

3.

When the desired number is reached,release the START button to startmeasurement. Continue to measure
your blood pressure as described on
the previous page.
English 11

Press and hold the button

START

Release the button
at the desired
systolic pressure
Refer to the previous pagefor measurement

Notes for Accurate Measurement
Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five to ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.
Should the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. Refer to page 5 for thedescription of symbols.
This blood pressure monitor is intended for use by adults only. Consult
with your physician before using this device on a child. A child should not
use this device unattended.

TriCheck Measurement
The TriCheck mode takes three consecutive measurements automaticallyand displays the average values of the three measurements.

Selecting the TriCheck ModeIn the pressure setting mode,1.

In the pressure setting mode, press
the SET button to enter the TriChecksetting mode. The current setting
blinks.

2.

Press theorbutton to turn the
TriCheck mode ON or OFF.
The current setting blinks
ON: TriCheck mode
OFF: Normal measurement mode (default value)or

3.

Press the START or SET button toturn the device off. After threeminutes of non-operation, the device
will turn off automatically.
English 12

press

Change the setting
or START

Measurement Using the TriCheck Mode1.

2.

Press the START button.
All of the display segments are
displayed.
Zero (0) is displayed blinking briefly
and the first measurement starts.
When the measurement is complete,
the systolic and diastolic pressure
readings and pulse rate are displayed,then a one-minute countdown starts forthe second measurement.

3.

After one minute, the second
measurement starts.

4.

When the measurement is complete,
the average readings of the first andsecond measurements are displayed,then a one-minute countdown starts forthe third measurement.

5.
6.

At heart level
START

Press
All of the display
segments displayed
First measurement

One minute after

After one minute, the third
measurement starts.

Results of thefirst measurement

Second measurement

When the measurement is complete,
the average readings of the threemeasurements are displayed andstored in memory.

Notes:
During the measurement, theTriCheck symbol is displayed.
To cancel the measurement, press
the START button. In this case, nodata is stored in memory.
When the cuff fit error symbol appears
during the first measurement, cancel
the measurement, reapply the cuff
correctly and start a new measurement.
After measurement, the averagedata of the three measurements isstored in memory.
No data will be stored, whencancelingthemeasurements,
before completion of the third
measurement.
English 13

One minute after

Average of the first andsecond measurements

Third measurement

Average of thethree measurements

Recalling the Memory Data
Note: This device stores the last 90 measurements in memory.
The device announces the memory data values as they are displayed.
or button.
1. Press the
The average of all measurements andthe number of data are displayed.
(If no data, “0” is displayed. Press the, or START button to turn thedevice off.)Press

2. Each time thebutton (or the
button to display the data in thereverse order) is pressed, the memory
data is displayed as follows.
Average of all AM (morning) measurements
taken between 4:00 and 9:59.
(In the example, 10 measurements. If
no data, “--” is displayed.)or
Average systolic
Average diastolic
Average pulse

Average of
all the data

Averageof AM data

Average of all PM (evening) measurements
taken between 18:00 and 1:59.
(In the example, 9 measurements. If
no data, “--” is displayed.)
Most recent data (No.n, in theexample, No.35)
Three seconds after the data number
display, the measurement data is displayed.

Last data (No.1)
Three seconds after the data number
display, the measurement data isdisplayed.

Average systolic
Average diastolic
Average pulse

Average systolic
Average diastolic
Average pulse
Averageof PM data
Systolic
Diastolic
Pulse
Most recent data

3. After the last data is displayed, press
thebutton to return the average
display of all measurements.
4. Press the START button to turn thedevice off. After one minute of
non-operation, the device will turn off
automatically.
English 14

Systolic
Diastolic
Pulse
Last data (Oldest)Adjusting the Speaker Volume
1. Press the VOLUME button to enter thevolume adjustment mode.
Six stages of volume, which are
represented by bars, and OFF areavailable.

High

2.

Low

3.

Press thebutton to increase thevolume, or press the
button to
decrease the volume.
Press the START button to turn thedevice off. After three minutes of
non-operation, the device will turn off
automatically.

Press

Volume off
The current setting appearsorChange the volume

Selecting the Language
1.

2.

In the volume adjustment mode, press
the VOLUME button to enter the
language selection mode. The current
setting appears.

press

or
button to select a
Press thelanguage from the following.
L1: English (default value)
L2: French
L3: Spanish
L4: Italian
L5: Chinese

3.

In the volume adjustment mode,

The current setting appearsorChange the language

Press the START or VOLUME button
to turn the device off. After threeminutes of non-operation, the device
will turn off automatically.

English 15

What is an Irregular Heartbeat
The UA-1030T blood pressure monitor provides a blood pressure andpulse rate measurement even when an irregular heartbeat occurs. An
irregular heartbeat is defined as a heartbeat that varies by 25% from theaverage of all heartbeats during the blood pressure measurement. It is
important that you are relaxed, remain still and do not talk during
measurements.
Note: We recommend contacting your physician if you see thissymbol frequently.

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing

Releasing air

Inflation in progress

Measurement in progress
Inflation complete

WHO Classification Indicator
Each segment of the bar indicator
corresponds to the WHO blood pressureclassification described on the next page.
WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimalExample:Moderate hypertension

Mild hypertension: The indicator displays a segment, basedon the current data, corresponding to the
WHO classification.
High normal
English 16

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of the arteries.
Systolic pressure occurs when the heart contracts. Diastolic pressure occurs
when the heart expands. Blood pressure is measured in millimeters of
mercury (mmHg). One's natural blood pressure is represented by thefundamental pressure, which is measured first thing in the morning while one
is still at rest and before eating.

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left unattended,
can cause many health problems including stroke and heart attack.
Hypertension can be controlled by altering lifestyle, avoiding stress, and withmedication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?
Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

WHO Blood Pressure Classification
Standards to assess highbloodpressure,
without
regard to age, have been
established by the World
Health Organization (WHO),as shown in the chart.

Blood Pressure Variations
An individual’s blood pressure
varies greatly on a daily andseasonal basis. It may vary by
30 to 50 mmHg due to various conditions during the day. In hypertensive
individuals, variations are even more pronounced. Normally, the bloodpressure rises while at work or play and falls to its lowest levels during sleep.
So, do not be overly concerned by the results of one measurement.
Take measurements at the same time every day using the procedure
English 17

described in this manual to get to know
your normal blood pressure. Regularreadings give a more comprehensive bloodpressure history. Be sure to note the date
and time when recording your bloodpressure. Consult your doctor to interpret
your blood pressure data.

Troubleshooting
ProblemPossible Reason
Nothing
Batteries are drained.
appears on thedisplay, even
Battery terminals are notwhen the power in the correct position.
is turned on.
Battery voltage is too low.
(LOW BATTERY mark)The cuff does
blinks. If the batteries are
not inflate.
drained completely, themark does not appear.
The cuff is not applied
properly.
You moved your arm or
body during
The device does measurement.
not measure.
Readings are
The cuff position is not
too high or too correct.
low.

Other

The value is different
from that measured at aclinic or doctor’s office.

Recommended Action
Replace all batteries with new ones.
Reinstall the batteries withnegative and positive terminals
matching those indicated on thebattery compartment.
Replace all batteries with new
ones.
Apply the cuff correctly.
Make sure you remain still andquiet during measurement.
Sit comfortably and still.
Place your arm on a table with yourpalm facing upward and the cuff at
the same level as your heart.
If you have a very weak or
irregular heat beat, the device
may have difficulty in determining
your blood pressure.
Refer to “Why Measure Blood
Pressure at Home?”.

Remove the batteries. Placethem back properly and takeanother measurement.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this product, as any attempt todo so will make your warranty invalid.
English 18

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem using
the troubleshooting instructions, contact the authorized dealer in your areaor our customer service department. The A&D customer service will
provide technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life.
However it is generally recommended to have the device inspected every
2 years, to ensure proper functioning and accuracy. Please contact theauthorized dealer in your area or A&D for maintenance.

Technical Data
Type

UA-1030T

Measurement method

Oscillometric measurement

Measurement range

Pressure:
0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / minute

Measurement accuracy

Pressure:
Pulse:

Power supply

4 x 1.5V batteries (R6P, LR6 or AA) or
AC adapter (TB-233) (Not included)

Number of measurements

Approx. 200 measurements, when AA manganesebatteries are used, with pressure value of 180
mmHg at room temperature of 23°C

Classification

Internally powered ME equipment (Supplied bybatteries) /Class II (Supplied by adapter)
Continuous operation mode

Clinical test

According to ANSI / AAMI SP-10 1992

EMC

IEC 60601-1-2: 2007

MemoryLast 90 measurements

Operating conditions+10°C to +40°C / 15%RH to 85%RH
800 hPa to 1060 hPa±3 mmHg±5%

Transport / Storage conditions -20°C to +60°C / 10%RH to 95%RH
Dimensions

Approx. 140 [W] x 60 [H] x 105 [D] mm
English 19

Weight

Approx. 300 g, excluding the batteries

Applied part
Useful life

Cuff Type BF
Device: 5 years (when used six times a day)Cuff: 2 years (when used six times a day)Accessory AC adapter

The adapter is to connect the blood pressure
monitor to a power source at home.

TB-233

Please contact your local A&D dealer forpurchasing.
The AC adapter is required to be inspected or
replaced periodically.
TB-233C

Input: 100-240V
Output: 6V
139°C

TB-233BF

Input: 240V
Output: 6V

500mA
2A

500mA
139°C

2A

Accessories sold separately
Cuff
Catalog Number
CUF-F-LA
CUF-G-A

Cuff Size
Arm Size
Large adult cuff 31 cm to 45 cm
Adult cuff
17 cm to 32 cm

Catalog Number
TB-233C
TB-233BF

Plug
Type CType BF

AC adapter

Note: Specifications are subject to change without prior notice.
EMC table information is listed on our website:
http://www.aandd.jp/products/manual/medical/emc_acadapter_en.pdf

English 20

Asahi 1-243, Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
OX14 1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119
A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409