Digital Blood Pressure Monitor
Model UA-767S
Instruction Manual Original

EnglishManuel d’instructions Traduction

Français

Manual de Instrucciones Traducción

EspañolManuale di Istruzioni Traduzione

Italiano

Manual de instruções Tradução

Português

Instructiehandleiding Vertaling

Nederlands

Betjeningsvejledning Oversættelse Dansk
Bruksanvisning Översättning

Svenska

Bruksanvisning Oversettelse

Norsk

Ohjekirja Käännös

1WMPD4002781K

Suomi

Talimat Kılavuzu ÇeviriTÜRKÇE‫ﺗﺭﺟﻣﻪ ﺩﻟﻳﻝ ﺍﻹﺳﺗﻌﻣﺎﻝ‬‫ﺗﺮﺟﻤﻪ ﺩﻓﺘﺮﭼﻪ ﺭﺍﻫﻨﻤﺎ‬‫ﻋﺭﺑﻲ‬‫ﻓﺎﺭﺳﯽ‬Contents
Dear Customers .................................................................... 2
Preliminary Remarks ............................................................. 2
Precautions ........................................................................... 2
Parts Identification ................................................................ 5
Symbols ................................................................................ 6
Operation Mode .................................................................... 8
Using the Monitor ................................................................. 9
Installing / Changing the Batteries ................................... 9Connecting the Air Hose ................................................ 10
Connecting the AC Adapter ............................................ 10
Selecting the Correct Cuff Size ....................................... 11
Applying the Arm Cuff ................................................... 11
How to Take Accurate Measurements ............................. 13
Measurement ................................................................. 13
After Measurement ........................................................ 13
Measurements .................................................................... 14
Normal Measurement ..................................................... 14
Measurement with the Desired Systolic Pressure ............ 15
Notes for Accurate Measurement ................................... 15
Recalling the Memory Data ................................................. 16
What Is The IHB/AFib Indicator? .......................................... 17
What Is The AFib? ............................................................... 17％IHB/AFib .......................................................................... 17
Pressure Bar Indicator ......................................................... 18
WHO Classification Indicator ............................................... 18
About Blood Pressure.......................................................... 18
What is Blood Pressure? ................................................. 18
What is Hypertension and How is it Controlled? .............. 19
Why Measure Blood Pressure at Home?........................... 19
WHO Blood Pressure Classification ................................. 19
Blood Pressure Variations............................................... 19
Troubleshooting ................................................................. 20
Maintenance ....................................................................... 21
Technical Data .................................................................... 21

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure
monitor. Designed for ease of use and accuracy, this device will facilitate
your daily blood pressure regimen.
We recommend that you read through this manual carefully before
using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
mark of conformity. (0123:
Products. This is made evident by the
The reference number to the involved notified body)
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in thehome healthcare environment.
This device is designed to measure blood pressure and pulse rate ofpeople for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremesin temperature, humidity, direct sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff or storing the hose tightly twisted for long
periods, as such treatment may shorten the life of the components.
Take care to avoid accidental strangulation of babies or infants with the hoseand cable.
Do not twist the air hose during measurement. This may cause injury due
to continuous cuff pressure.
The device and cuff are not water resistant. Prevent rain, sweat and waterfrom soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
Wireless communication devices, such as home networking devices,mobile phones, cordless phones and their base stations, walkie-talkies canaffect this blood pressure monitor. Therefore, a minimum distance of 30
cm should be kept from such devices.
English 2

When reusing the device, confirm that the device is clean.
Used equipment, parts and batteries are not treated as ordinary
household waste, and must be disposed of according to the applicable
local regulations.
When the AC adapter is used, make sure that the AC adapter can bereadily removed from the electrical outlet when necessary.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the arm must be squeezed by the cuff hardenough to temporarily stop blood flow through the artery. This may cause
pain, numbness or a temporary red mark to the arm. This condition willappear especially when measurement is repeated successively. Any pain,
numbness, or red marks will disappear with time.
Measuring blood pressure too frequently may cause harm due to bloodflow interference. Check that the operation of the device does not result in
prolonged impairment of blood circulation, when using the device
repeatedly.
If you have had a mastectomy, please consult a doctor before using thedevice.
Do not let children use the device by themselves and do not use the device
in a place within the reach of infants. It may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
Unplug the AC adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially causeburns.
Allow the device to adapt to the surrounding environment before use (about
one hour).
Clinical testing has not been conducted on newborn infants and pregnant
woman. Do not use on newborn infants or pregnant woman.
Do not touch the batteries, the DC jack, and the patient at the same time.
That may result in electrical shock.
Do not inflate without wrapping the cuff around the upper arm.

Contraindications
The following are precautions for proper use of the device.
Do not apply the cuff on an arm with another medical electrical equipment
attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult adoctor before using the device, to avoid medical problems.
Do not self-diagnose the measurement results and start treatment by
yourself. Always consult your doctor for evaluation of the results andtreatment.
Do not apply the cuff on an arm with an unhealed wound.
English 3

Do not apply the cuff on an arm receiving an intravenous drip or bloodtransfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, suchas a high-pressure oxygen chamber or an oxygen tent. It may cause a fireor explosion.

English 4

Parts Identification
Display

Air Connector Plug

AC adapter Plug
DC Jack
START Button

Air Socket

Air Hose

Proper Fit Range

Artery Position Mark
Index MarkArm Cuff
Battery Compartment
Battery Cover

1.5V Batteries(R6P, LR6 or AA)

Display

MEMORY

Average

IHB/AFib symbol
Systolic
Pressure%IHB/AFib
Cuff Fit Error Symbol

Diastolic
Pressure

Movement Error Symbol
Heart MarkWHO Classification Indicatorand Pressure Bar Indicator

Pulse Rate

Battery Indicator
English 5

Symbols
Symbols that are printed on the device caseSymbols
SYS
DIA
PUL

IP

Function / MeaningStandby and Turn the device on.
Systolic blood pressure in mmHg
Diastolic blood pressure in mmHg
Pulse per minuteBattery installation guide
Direct current
Type BF: Device, cuff and tubing are designed to provide
special protection against electrical shocks.
EC directive medical device label
EU-representative
Manufacturer
Date of manufactureInternational protection symbol
WEEE label

SN

Serial number
Refer to instruction manual/booklet
Polarity of DC jack
Keep drySymbols that appear on the display
Symbols

Function / Meaning
Recommended Action
Appears while measurement is in
Measurement is inprogress. It blinks when the pulse isprogress. Remain as
detected.
still as possible.
IHB/AFib symbol
Appears when an irregular heartbeat
is detected.
It may light when a very slight vibration
like shivering or shaking is detected.
The reading may yield
an incorrect value.
Appears when a body or arm movement
Try the measurement
is detected.
again. Remain still
during measurement.
English 6

Symbols
Symbols

Function / Meaning
Appears during measurement
when the cuff is attached loosely

Recommended Action
The reading may yield anincorrect value. Apply the
cuff correctly and try themeasurement again.

Detected rate of IHB/AFib in memory%IHB/AFib =Number of detected
IHB/AFibs in memory

x100 [%]

Total number

Previous measurements stored in
MEMORY.
Average data
FULL BATTERY
The battery power indicator during
measurement.
LOW BATTERY
Replace all batteries with newones when the mark blinks.
The battery is low when it blinks.
Try the measurement again.
Unstable blood pressure due toRemain very still during
movement during measurement.
measurement.
The systolic and diastolic values
are within 10 mmHg of each other.
The pressure value did not
Apply the cuff correctly,
increase during inflation.
and try the measurement
again.
The cuff is not applied correctly.

PUL DISPLAY ERROR
The pulse is not detected correctly.
Blood pressure monitor internal
errorEnglish 7

Remove the batteries andpress the START button,
and then install the batteries
again. If the error still
appears, contact the dealer.

Operation Mode
1. Normal Measurement
Press the START button. Blood pressure is measured and the data is stored
in memory. This device can store the last 60 measurements in memory.

2. Recalling the Data
When nothing is displayed, press and holdthe START button.

In standby,press and hold the
STARTbuttonRelease the button when displaying the
average data.

The data number and stored data areautomatically displayed in order from the lastmeasurement.

For details on recalling the data, Refer to“Recalling the Memory Data”.

3. Deleting all Data Stored in Memory
When turning off the device, press and hold the
START button until the "" is displayed.
Select "" to clear the data.
The data is cleared when themark blinks.
The device turns off automatically.

START

4. Measurement with the Desired Systolic Pressure
Refer to page 15 for measurement with the desired systolic pressure.

English 8

Using the Monitor
Installing / Changing the Batteries
1. Remove the battery cover.
2. Remove the used batteries andinsert new batteries into the battery
compartment as shown, taking
care that the polarities (+ and -) arecorrect.
Use only R6P, LR6 or AA batteries.
3. Attach the battery cover.

Step 1
Step 3

Step 1
Step 2

CAUTION
Insert the batteries as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(LOW BATTERY mark) blinks in the display, replace allbatteries with new ones. Do not mix old and new batteries. It may shorten
the battery life, or cause the device to malfunction.
Replace the batteries two seconds or more after the device turns off.
If
(LOW BATTERY mark) appears even after the batteries are replaced,make a blood pressure measurement. The device may then recognize the newbatteries.
(LOW BATTERY mark) does not appear when the batteries are
drained.
The battery life varies with the ambient temperature and may be shorter at
low temperatures. Generally, four new LR6 batteries will last approximately
for one year when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device arefor testing monitor performance and may have a limited life.
Remove the batteries if the device is not to be used for a long time.
The batteries may leak and cause a malfunction.

English 9

Using the Monitor
Connecting the Air Hose

Air socket

Insert the air connector plug into
the air socket firmly.

Air connector plug

Connecting the AC Adapter
Insert the AC adapter plug into the
DC jack.
Next, connect the AC adapter to anelectrical outlet.
Use the specified AC adapter.
(Refer to page 22.)

DC jack
AC adapter plug

When disconnecting the AC adapter from the electrical outlet, grasp
and pull the AC adapter body out of the outlet.
When disconnecting the AC adapter plug from the blood pressure
monitor, grasp and pull the AC adapter plug out of the monitor.

English 10

Using the Monitor
Selecting the Correct Cuff Size
Using the correct cuff size is important for an accurate reading. If the cuff is
not the proper size, the reading may yield an incorrect blood pressure value.
The arm size is printed on each cuff.
The index
and proper fit range, on the cuff, tell you if you are applying thecorrect cuff. (Refer to "Symbols that are printed on the cuff" on the next page.)
If the index
points outside of the range, contact your local dealer topurchase a replacement cuff.
The arm cuff is a consumable. If it becomes worn, purchase a new one.
Arm Size
Recommended Cuff Size
Catalog Number
31 cm to 45 cm
Large Adult Cuff
CUF-F-LA
22 cm to 42 cm
Wide Range Cuff
CUF-I
22 cm to 32 cm
Adult Cuff
CUF-F-A
Arm size: The circumference at the biceps.
Note:
The UA-767S is not designed for using a small cuff.

Applying the Arm Cuff
1. Wrap the cuff around the upper arm,
about 1-2 cm above the inside of theelbow, as shown.
Place the cuff directly against theskin, as clothing may cause a faintpulse, and result in a measurement
error.

Do not roll up shirtsleeve tightly.
Cuff

Air hose
Artery position mark2. Constriction of the upper arm,
caused by tightly rolling up a
shirtsleeve, may prevent accurate
readings.
3. Confirm that the index
within the proper fit range.

1-2 cmSecure the fabricfastener firmly.

pointsIndex

English 11

and proper fit range

Using the Monitor
Symbols that are printed on the cuff
Symbols
Function / Meaning

Recommended Action
mark on the
Set theartery of the upper arm orin line with the ring finger
on the inside of the arm.

Artery position mark
▲
REF

Index
Catalog number
Proper fit range for the adult cuff.
It's printed on the adult cuff.
Use the large adult cuff
Over range printed on the adult cuff
instead of the adult cuff orand wide range cuff.
wide range cuff.
Proper fit range for the wide range cuff.
It's printed on the wide range cuff.
Proper fit range for the large adult cuff.
It's printed on the large adult cuff.
Under range printed on the adult cuff
and wide range cuff.
Use the adult cuff instead
Under range on the large adult cuff.
of the large adult cuff.
Lot number

Large adult cuff

Proper fit range

A

L

Wide range cuff

S

W

L

A

L

Adult cuffS

English 12

Using the Monitor
How to Take Accurate Measurements
For the most accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross yourlegs. Keep your feet flat on the floor and straighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same level as your heart.
Remain still and keep quiet during measurement.
Do not measure right after physical exercise or a bath. Rest for twenty or
thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

Measurement
During measurement, it is normal for the cuff to feel very tight. (Do not be
alarmed.)After Measurement
After measurement, press the START button to turn the device off.
Remove the cuff and record your data.
Note: The device has an automatic power shut-off function, which turns thedevice off approximately one minute after measurement.
Allow at least three minutes between measurements on the same person.

English 13

Measurements
Before measurement, refer to “Notes for Accurate Measurement” on the next
page.

Normal Measurement
1.
2.

Place the cuff on the arm (preferably the left arm).
Sit quietly during measurement.

At heart level

Press the START button.
All of the display segments are displayed.
Press
Zero (0) is displayed blinking briefly.
The display changes, as indicated in the
All of the display
figure at the right, as the measurement
begins. The cuff starts to inflate. It is segments displayed
normal for the cuff to feel very tight. A
pressure bar indicator is displayed, on theleft edge of the display, during inflation.
Zero display
Note: If you wish to stop inflation at any time,
Starts inflationpress the START button again.

3.

When inflation is complete, deflation starts
automatically and(heart mark) blinks,
indicating that the measurement is in
Pressurizing
progress. Once the pulse is detected, themark blinks with each pulse beat.
Note: If an appropriate pressure is not obtained, thedevice starts to inflate again automatically.
To avoid re-inflation, refer to “Measurementwith the Desired Systolic Pressure” on thenext page.

4.

When the measurement is complete, thesystolic and diastolic pressure readings
and pulse rate are displayed.
The cuff exhausts the remaining air anddeflates completely.

5.

Press the START button to turn the device off.

Note: Allow at least three minutes between
measurements on the same person.

START

Measurementin progress
Systolic pressure
Diastolic pressure
WHO classification
Pulse rate
Exhausts remaining
air automaticallyEnglish 14

Measurements
Measurement with the Desired Systolic Pressure
The UA-767S is designed to detect the pulse and to inflate the cuff to asystolic pressure level automatically.
Use this method when re-inflation occurs repeatedly or when the results
are not displayed even if the pressure decreases to 20 mmHg or less.
1.

Place the cuff on the arm at heart level(preferably the left arm).

At heart level2.

Press the START button.

Press the START
button3.

When the zero blinks, press and holdthe START button until a number
about 30 to 40 mmHg higher thanyour expected systolic pressure
appears.

4.

When the desired number is reached,release the START button to startmeasurement. Continue to measure
your blood pressure as described on
the previous page.

Release the button
at the desired
systolic pressure

Press and hold the
buttonSTARTRefer to the previous pagefor measurement

Notes for Accurate Measurement
Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five to ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.
Should the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. Refer to page 7 for thedescription of symbols.
English 15

This blood pressure monitor is intended for use by adults. Consult withyour physician before using this device on a child. A child should not use
this device unattended.
The automatic blood pressure monitor’s performance may be affected byexcessive temperature or humidity, or altitude.

Recalling the Memory Data
Note: This device stores the last 60 measurements in memory.
1. When nothing is displayed, press
and hold the _START_ button torecall the stored data.

Press and START button
hold
Average systolic
Average diastolic

2. Release the button when displaying
the average data.

Average pulse
Average of
all the data

3. The data number and stored data
are automatically displayed in orderfrom the last measurement.
Systolic
4. The display will turn off automaticallyafter all data are displayed.

Diastolic
Pulse
Most recent data

Note: If you press the _START_ button
while recalling data, the device
turns off.

Systolic
Diastolic
Pulse
Last data (Oldest)

English 16

What Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements,
the IHB/AFib indicator will appear on the display with the measurement
values.
Note: We recommend contacting your physician if you see thisIHB/AFib indicator frequently.

What Is The AFib?
The heart contracts due to electrical signals occurring in heart and sends
blood through the body. Atrial fibrillation (AFib) occurs when the electricalsignal in the atrium becomes confused and leads to disturbances in thepulse interval. AFib can cause blood to stagnate in the heart, which caneasily create clots of blood, a cause of stroke and heart attack.

％IHB/AFib%IHB/AFib is displayed as frequency of IHB/AFib detected.
IHB/AFib can detect not only noises such as physical movement but also
an irregular heartbeat. Therefore, we recommend contacting yourphysician if %IHB/AFib level is high.
Number of detected
IHB/AFibs in memory
x 100 [%]
%IHB/AFib =Total number
Display of %IHB/AFib: %IHB/AFib is displayed when displaying average values.
(Refer to “2. Recalling the Data” in “Operation Mode”)%IHB/AFib is not displayed when the memory number is six or less.
Average value display

Level 0
%IHB/AFib=0

Level 1
%IHB/AFib=1 - 9

Level 2
%IHB/AFib=10 - 24

Not displayed

English 17

Level 3
%IHB/AFib=25 - 100

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing

Releasing air

Inflation in progress

Measurement in progress
Inflation complete

WHO Classification Indicator
Each segment of the bar indicator
corresponds to the WHO blood pressureclassification described on the next page.
WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimal: The indicator displays a segment, basedon the current data, corresponding to the
WHO classification.

Example:Moderate hypertension

Mild hypertension

High normal

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of thearteries. Systolic pressure occurs when the heart contracts. Diastolicpressure occurs when the heart expands. Blood pressure is measured inmillimeters of mercury (mmHg). One's natural blood pressure isrepresented by the fundamental pressure, which is measured first thing in
the morning while one is still at rest and before eating.
English 18

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left
unattended, can cause many health problems including stroke and heart
attack. Hypertension can be controlled by altering lifestyle, avoiding
stress, and with medication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?
Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

WHO Blood Pressure Classification

Blood Pressure Variations

Diastolic blood pressureStandards to assess high Reference Material: Journal of Hypertension 1999, Vol 17 No.2
Grade 3 hypertension (severe)bloodpressure,
without mmHg
110
regard to age, have been
Grade 2 hypertension (moderate)105
established by the World
100
Health Organization (WHO),
Grade 1 hypertension (mild)95
as shown in the chart.
90
85
80

High-normal

NormalOptimal120 130 140 150 160 170 180
Systolic blood pressure
mmHg

Blood pressureAn
individual’s
bloodpressure varies greatly on adaily and seasonal basis. It
may vary by 30 to 50 mmHg due to various conditions during the day. In
hypertensive individuals, variations are even more pronounced. Normally,
the blood pressure rises while at work or play and falls to its lowest levels
during sleep. So, do not be overly concerned by the results of one
measurement.
Typical fluctuation within a day
Take measurements at the same time every(Measured every five minutes)
day using the procedure described in thisSystolic
manual to get to know your normal bloodpressure. Regular readings give a morecomprehensive blood pressure history. Be
sure to note the date and time whenrecording your blood pressure. Consult yourSleep
Diastolicdoctor to interpret your blood pressure data.
PM

English 19

AM

Time

Troubleshooting
ProblemPossible Reason

Recommended Action
Replace all batteries with newBatteries are drained.
Nothing
ones.
appears in the
Reinstall the batteries withdisplay, even
Battery terminals are not negative and positive terminals
when the device
in the correct position.
matching those indicated on theis turned on.
battery compartment.
Battery voltage is too low.
(LOW BATTERY mark)The cuff does
Replace all batteries with new
blinks. If the batteries are
not inflate.
ones.
drained completely, themark does not appear.
The cuff is not applied
Apply the cuff correctly.
properly.
You moved your arm or
Make sure you remain very stillbody during measurement. and quiet during measurement.
The device does
Sit comfortably and still.
not measure.
The cuff position is not Place your arm on a table with yourReadings arecorrect.
palm facing upward and the cuff at
too high or too
the same level as your heart.
low.
If you have a very weak or
irregular heartbeat, the device
may have difficulty in determining
your blood pressure.
The value is different
Refer to “Why Measure Bloodfrom that measured at a
Pressure at Home?”.
clinic or doctor’s office.
Other
Remove the batteries. Placethem back properly and try themeasurement again.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this product, as any attempt todo so will make your warranty invalid.

English 20

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem using
the troubleshooting instructions, contact the authorized dealer in your areaor our customer service department. The A&D customer service will
provide technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life.
However it is generally recommended to have the device inspected every
2 years, to ensure proper functioning and accuracy. Please contact theauthorized dealer in your area or A&D for maintenance.

Technical DataType
Measurement method
Measurement range

UA-767S
Oscillometric measurement
Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / minute
Measurement accuracy
Pressure: ±3 mmHg
Pulse:±5%
Power supply
4 x 1.5V batteries (R6P, LR6 or AA) or
AC adapter (TB-233C) (Not included)
Number of measurements
Approx. 700 times LR6 (alkaline batteries)
Approx. 200 times R6P (manganese batteries)
With pressure value of 180 mmHg at room
temperature of 23 °C.
Classification
Internally powered ME equipment (Supplied bybatteries) / Class II (Supplied by adapter)
Continuous operation mode
Clinical test
According to ISO81060-2：2013
In the clinical validation study, K5 was used on 85
subjects for determination of diastolic blood pressure.
EMD
IEC 60601-1-2: 2014
MemoryLast 60 measurements
Operating conditions+10 to +40 °C / 15 to 85 %RH / 800 to 1060 hPa
Transport / Storage conditions -20 to +60 °C / 10 to 95 %RH / 700 to 1060 hPa
Dimensions
Approx. 140 [W] x 60 [H] x 105 [D] mm
WeightApprox. 245 g, excluding the batteries
English 21

Ingress protection
Applied part

Device:Cuff

IP20
Type BF

Useful life

Device:5 years (when used six times a day)Cuff:
2 years (when used six times a day)AC adapter: 5 years (when used six times a day)Accessory AC adapter
The adapter is to connect the device to a power source at home. Pleasecontact your local A&D dealer for purchasing. The AC adapter is required to
be inspected or replaced periodically.
TB-233C
Please contact your local A&D dealer forpurchasing.
The AC adapter is required to be inspected or
replaced periodically.
Symbols that are printed on the AC adapter
Symbols
Function / Meaning
For indoor use only
Class II device
Thermal fuse
Fuse
EC directive device label
EAC certification device label
Polarity of AC adapter plug
Accessories sold separately
Cuff
Catalog Number
Cuff Size
Arm Size
CUF-F-LA
Large adult cuff
31 cm to 45 cm
CUF-I
Wide range cuff
22 cm to 42 cm
CUF-F-A
Adult cuff
22 cm to 32 cm
Arm size: The circumference at the biceps.
AC adapter

Catalog Number
TB-233C

Plug (Outlet type)Type C

Note: Specifications are subject to change without prior notice.
IP classification is the degrees of protection provided by enclosures inaccordance with IEC 60529. This device is protected against solid
foreign objects of 12 mm diameter and greater such as a fingers. This
device is not protected against water.

English 22

Sommaires
Chers clients ......................................................................... 2
Remarques préliminaires ...................................................... 2
Précautions ........................................................................... 2
Nomenclature ....................................................................... 5
Symboles .............................................................................. 6
Mode opératoire ................................................................... 8
Utilisation de l’appareil ......................................................... 9Mise en place / Remplacement des piles .......................... 9
Branchement du tuyau d’air ........................................... 10
Branchement de l’adaptateur secteur ............................. 10
Choix du brassard ......................................................... 11
Mise en place du brassard .............................................. 11
Comment faire des mesures correctes .............................. 13
Pendant la mesure ......................................................... 13
Après la mesure ............................................................. 13
Les mesures ....................................................................... 14
Mesure normale ............................................................. 14
Mesure avec la pression systolique désirée .................... 15
Consignes et observations pour une mesure correcte..... 15
Rappel des données mémorisées ........................................ 16
Qu’est ce que l’indicateur de RCI/FA? ................................. 17
Qu’est-ce que la FA? ........................................................... 17％RCI/FA ............................................................................. 17
Indicateur de progression du gonflage ................................ 18
Indicateur de tension selon la classification OMS ......................... 18
Pression sanguine ............................................................... 18
Qu'est-ce que la pression sanguine? .............................. 18
Qu'est-ce que l'hypertension et comment la contrôler? .. 19
Comment mesurer la pression sanguine chez soi?.......... 19
Classification de la tension selon l’OMS ......................... 19
Variations de la pression sanguine ................................. 19
Résolution des problèmes ................................................... 20
Entretien ............................................................................. 21
Fiche technique .................................................................. 21

Français

1

Chers clients
Félicitations pour l’achat de ce moniteur de la pression sanguine A&D derniercri. Etant étudié pour faciliter l’utilisation et la précision, ce moniteur vous
donnera au quotidien des mesures précises de la pression sanguine.
Lisez ce manuel entièrement et soigneusement avant d’utiliser l’appareil
pour la première fois.

Remarques préliminaires
Cet appareil est conforme à la directive européenne 93/42 EEC des appareils
médicaux avec le marquagede conformité. (0123: Le numéro deréférence au corps notifié concerné)
Cet appareil est conçu pour être utilisé sur les adultes, et non sur lesnouveau-nés ou les enfants.
Lieu d’utilisation. Cet appareil est destiné à être utilisé par vous-même dans unenvironnement domestique de soins de santé.
Cet appareil est conçu pour mesurer la tension artérielle et la fréquencecardiaque des personnes dans un objectif de diagnostic.

Précautions
Des composants de précision sont utilisés dans la construction de cet appareil.
Il convient d'éviter les extrêmes en matière de température, d'humidité,d'exposition à la lumière directe du soleil, de chocs ou de poussière.
Nettoyez l'appareil et le brassard à l'aide d'un chiffon sec et doux ou d'unchiffon imbibé d'eau et d'un détergent neutre. N'utilisez jamais d'alcool, debenzène, de diluant ou d'autres produits chimiques agressifs pour nettoyer
l'appareil ou le brassard.
Évitez de plier fortement le brassard ou de ranger le tuyau fortement tordu
pendant des périodes prolongées, car cela pourrait réduire la durée de vie
des accessoires.
Veillez à ce que les bébés et les enfants en bas âge ne s’étranglent pas avec
le tuyau et le câble.
Ne tordez pas le tuyau d’air pendant la mesure. La pression continue dubrassard pourrait vous blesser.
L'appareil et le brassard ne sont pas étanches. Évitez que la pluie, la sueur ou
de l'eau ne souillent l'appareil et le brassard.
Il est possible que les mesures soient altérées si l’appareil est utilisé à proximité
de téléviseurs, de fours à micro-ondes, de téléphones cellulaires, de sources derayonnement X ou d’autres appareils émettant de forts champs électriques.
Les dispositifs de communication sans fil, tels que les dispositifs réseau à
domicile, les téléphones mobiles, les téléphones sans fil et leurs stations de base,les talkies-walkies peuvent affecter ce moniteur de pression sanguine. Une
distance minimale de 30 cm doit donc être observée pour ces dispositifs.
Si vous réutilisez l'appareil, vérifiez que celui-ci est propre.
Français2

L'équipement, les pièces et les piles usagées ne doivent pas être traités
comme des déchets ménagers ordinaires et doivent être mis au rebut
conformément aux règlementations locales applicables.
Lorsque vous utilisez l'adaptateur secteur, assurez-vous que celui-ci peut êtrefacilement retiré de la prise électrique si nécessaire.
Ne modifiez pas l'appareil. Cela pourrait provoquer des accidents ou
endommager l'appareil.
Pour mesurer la pression artérielle, le bras doit être comprimé de manière
suffisamment forte par le brassard pour arrêter temporairement la circulation
sanguine dans l'artère. Il est possible que cela provoque une douleur, unengourdissement ou une marque rouge temporaire sur le bras. Cet étatapparaît en particulier si la mesure est répétée à plusieurs reprises. La douleur,l'engourdissement ou les marques rouges disparaissent avec le temps.
Une mesure trop fréquente de la pression sanguine peut nuire au flux sanguin.
Si vous utilisez l’appareil à plusieurs reprises, assurez-vous que l’utilisation del’appareil n’entrave pas la circulation sanguine de manière prolongée.
Veuillez consulter un médecin avant d’utiliser cet appareil si vous avez subi
une mastectomie.
Ne laissez pas d’enfants utiliser l’appareil par eux-mêmes et n’utilisez pas l’appareildans un endroit où il se trouve à portée d’enfants en bas âge. Cela pourrait
provoquer des accidents ou des dommages.
L’appareil comporte de petites pièces qui risqueraient d’étouffer les enfants en bas
âge s’ils venaient à les avaler.
Débranchez l’adaptateur secteur lorsqu’il n’est pas utilisé pendant la mesure.
L’utilisation d’accessoires non décrits dans ce manuel peut compromettre la sécurité.
Si la pile est en court-circuit, elle peut devenir chaude et causer potentiellement
des brûlures.
Laissez l’appareil s’adapter à l’environnement ambiant avant utilisation (environ
une heure).
Aucun essai clinique n’a été mené sur les nouveau-nés et les femmes enceintes.
N’utilisez pas l’appareil sur des nouveau-nés ou des femmes enceintes.
Ne touchez pas les piles, la prise CC et le patient en même temps. Cela pourrait
entraîner un choc électrique.
Ne gonflez pas sans avoir enroulé le brassard autour de la partie supérieure du bras.

Contre-indications
Vous trouverez ci-dessous les précautions pour l'utilisation appropriée de cet appareil.
N'appliquez pas le brassard sur un bras auquel un autre équipement médicalélectrique est fixé. L'équipement pourrait ne pas fonctionner correctement.
Les personnes ayant de graves problèmes de circulation au niveau du bras
doivent consulter un médecin avant d'utiliser l'appareil afin d'éviter tout
problème médical.
Vous ne devez pas procéder à un autodiagnostic du résultat des mesures etcommencer un traitement de votre propre initiative. Consultez toujours unmédecin pour une évaluation des résultats et du traitement.
N'appliquez pas le brassard sur un bras comportant une blessure nonsoignée.
Français3

N'appliquez pas le brassard sur un bras recevant un goutte-à-goutte
intraveineux ou une transfusion sanguine. Cela pourrait provoquer une
blessure ou des accidents.
N'utilisez pas l'appareil en présence de gaz inflammables tels que des gaz
anesthésiques. Cela pourrait provoquer une explosion.
N'utilisez pas l'appareil dans des environnements à forte concentration en
oxygène, tels qu'une chambre à oxygène haute pression ou une tente à
oxygène. Cela pourrait provoquer un incendie ou une explosion.

Français4Nomenclature
Affichage

Fiche del'adaptateur
secteur

Prise de connexion d’air

Connecteur decourant continu

Connecteur d’air

Bouton START(mise en marche)

Plage d’accord
appropriée

Tuyau d’air
Marque deposition de l’artère

Marque d’index

Brassard

Logement des piles

Couvercle logement des piles

Piles 1,5 V(R6P, LR6 ou AA)

Partie d’Affichage

MoyenneMEMOIRE
Symbole de RCI/FA

Pressionsystolique%RCI/FA

Symbole d’erreur de fixation du brassard

Pression
diastoliqueSymbole d’erreur de mouvement
Symbole du cœurIndicateur de la Classification
suivant l’OMS et Indicateur deprogression du gonflage

Pouls

Indicateur des piles
Français

5Symboles
Symboles imprimés sur l’appareil
SymbolesFonction / Signification
Pour allumer ou éteindre l’appareil

SYS

Pression systolique en mmHg.

DIA

Pression diastolique en mmHg.

PUL

Pouls: nombre de pulsations cardiaques par minute.
Repères pour installer les piles
Courant continu
T Type BF: L’appareil, le brassard et les tubes isolants ont été
conçus de manière à fournir une protection particulière contre leschocs électriques.
Dispositif Médical conforme à la directive européenne 93/42 CEE
Mandataire Européen
FabricantDate de fabrication

IP

Symbole de protection internationale
Déchets d’Equipements Electriques et Electroniques (DEEE)

SN

Numéro de série
Consultez le manuel/la brochure d'instructions
Polarité du connecteur de courant continu
Maintenir sec

Symboles affichés à l'affichage
SymbolesFonction / Signification

Action recommandée

Cet indicateur apparaît lorsque la mesure
L’appareil est en train deest en cours. Il clignote quand lesmesurer; restez calme.
pulsations cardiaques sont détectées.
Symbole de RCI/FA
Ce symbole apparaît en cas de détectiond’un rythme cardiaque irrégulier.
Il est susceptible de s’allumer lors de ladétection d’une très faible vibration telle
qu’un frissonnement ou un tremblement.
La lecture peut être à l’origine
d’une valeur erronée.
Symbole affiché lors de la détection
Recommencez la mesure.
d'un mouvement du corps ou du bras. Restez parfaitement
tranquille pendant la mesure.
Français

6Symboles
SymbolesFonction / Signification

Action recommandéeSymbole affiché pendant la mesure si
le brassard n’est pas assez serré.

La lecture peut être à l’origine
d’une valeur erronée. Attachez
le brassard correctement, etrecommencez la mesure.

Taux détecté de RCI/FA en mémoire
Nombre de RCI/FA%RCI/FA =détectés en mémoire

x 100 [%]

Nombre total
Les mesures précédentes conservées
dans la MEMOIRE.
Données moyennes
PILE PLEINE

Indicateur de tension des piles pendant
la mesure.
Remplacer toutes lespiles par des neuves
La tension des piles est faible quand ilquand cet indicateur
clignote.
clignote.
Recommencez laApparaît en cas de fluctuation de lamesure. Restezpression due à un mouvement pendantparfaitement tranquillela mesure.
pendant la mesure.
Apparaît si la difference entre lapression systolique(tension maxima) etla pression diastolique(tension minima)
est inférieure à 10 mmHg.
Apparaît lorsque la pression n’augmente Attachez le brassard
pas pendant le gonflage du brassard.
correctement, etApparaît lorsque le brassard n’est pas recommencez la mesure.
attaché correctement.
PILE FAIBLEERREUR D'AFFICHAGE DE POULS

La pression n’est pas détectéecorrectement.
Erreur interne du moniteur de lapression sanguine

Français

7

Retirez les piles et appuyez
sur le bouton START , puis
remettez les piles en place.
Si l’erreur s’affiche encore,contactez votre revendeur.

Mode opératoire
1. Mesure normale avec les données conservées
Appuyez sur le bouton START . La pression sanguine est mesurée et lesdonnées sont conservées dans l’appareil. Cet appareil peut conserver les
60 dernières données automatiquement dans la MEMOIRE.

2. Rappel des données en mémoire
Si rien ne s’affiche, appuyez de manière
prolongée sur le bouton START .

En veille,
appuyez de manière
prolongée sur le
START bouton

Relâchez le bouton lorsque les données
moyennes s'affichent.
Le numéro des données et les données
enregistrées s'affichent automatiquement dansl'ordre, à partir de la dernière mesure.
Pour plus de détails sur le rappel des données,
reportez-vous à la section « Rappel des donnéesmémorisées ».

3. Suppression de toutes les données en mémoire
Lorsque vous éteignez l’appareil, appuyez sur le bouton_START_ et maintenez-le enfoncé jusqu'à ce que«» s'affiche.
Sélectionnez «» pour effacer les données.
Les données sont effacées lorsque le symbole
clignote.
L'appareil s'éteint automatiquement.

START

4. Mesure avec la pression systolique désirée
Voir la page 15 pour la mesure avec la pression systolique désirée.
Français

8

Utilisation de l’appareil
Mise en place / Remplacement des piles
1. Enlevez le couvercle du logement despiles.

Etape 1

Etape 3

2. Enlevez des piles usées et insérez despiles neuves dans le logement en
respectant bien les polarités (+ et -)indiquées.
Utilisez uniquement des piles R6P,
LR6 ou AA.

Etape 1

Etape 2

3. Fermez le couvercle du logement des piles.
ATTENTION
Veillez à insérer les piles dans le bon sens, sinon l’appareil ne fonctionnera
pas.
Quand le symbole(PILE FAIBLE) clignote sur l’écran, l’appareil
indique que les piles doivent être remplacées. Changer alors toutes lespiles par des neuves. Ne mélangez pas les piles neuves et anciennes car
cela pourrait raccourcir leur durée de vie ou détériorer l’appareil.
Remplacez les piles au moins deux secondes après avoir éteint l'appareil.
Si le symbole(PILE FAIBLE) s'affiche même après le remplacement
des piles, réalisez une mesure de la pression sanguine. L'appareil devrait
alors reconnaître les nouvelles piles.
Le symbole(PILE FAIBLE) n'apparaît pas lorsque les piles sontépuisées.
La durée de vie des piles varie avec la température ambiante et peut êtreraccourcie par des températures basses. En général, quatre piles neuves
LR6 durent approximativement un an en cas de mesure deux fois par jour.
Utilisez uniquement les piles spécifiées. Les piles fournies avec l'appareil
sont pour tester les performances de l'écran d’affichage et peuvent donc
avoir une durée de vie plus courte.
Retirez les piles lorsque l'appareil n'est pas utilisé pendant un long moment.
Les piles pourraient couler ce qui entraînerait un mauvais fonctionnement
de l'appareil.

Français

9

Utilisation de l’appareil
Branchement du tuyau d’air

Connecteur d’air
Insérez fermement la prise deconnexion d’air dans le connecteur d’air.

Prise de connexion d’air

Branchement de l’adaptateur secteur
Introduire la fiche de l'adaptateur secteurdans le connecteur de courant continu.
Brancher ensuite l’adaptateur sur une prisede courant.

Connecteur de courant
continu
Fiche de l'adaptateur
secteur

Utilisez l'adaptateur secteur spécifié.
(Voir la page 22.)
Lorsque vous débranchez l’adaptateur secteur de la prise secteur, tenez
le corps de l’adaptateur et débranchez-le de la prise.
Lorsque vous débranchez l’adaptateur secteur du moniteur de pression
sanguine, tenez la fiche de l’adaptateur et débranchez-la du moniteur.

Français

10

Utilisation de l’appareil
Choix du brassard
L’utilisation d’un brassard adapté à la morphologie du patient est importante pourun résultat précis. Si la taille du brassard n’est pas appropriée il peut en résulterl’affichage de valeurs erronées.
La taille du brassard (circonférence du biceps) est imprimée sur chaque brassard.
L’indexet la plage d’accord appropriée sur le brassard vous indiquent si vous
mettez le brassard correctement. (Reportez-vous à « Symboles imprimés sur lebrassard », à la page suivante.)
Si l’index
se trouve en dehors de la plage, prière de prendre contact avec
votre revendeur local afin d’acheter le brassard pour le remplacement.
Le brassard est un produit consommable. S’il est usé, achetez-en un neuf.
Taille du bras
Taille du brassard recommandée Numéro du catalogue
31 cm à 45 cm
Brassard adulte de grande taille
CUF-F-LA
22 cm à 42 cm
Brassard de grande taille
CUF-I
22 cm à 32 cm
Brassard adulte
CUF-F-A
Taille du bras : circonférence au niveau du biceps.
Remarque :le modèle UA-767S n'est pas conçu pour une utilisation avec un
brassard adulte de petite taille.

Mise en place du brassard

Ne remontez pas la mancheétroitement.

1. Enroulez le brassard autour du bras,à 1 ou 2 cm au-dessus de l’intérieur
du coude, comme indiqué surl’illustration ci-contre. Placez lebrassard directement sur la peau,
car un vêtement pourrait assourdir
la pulsation et provoquer une erreurde mesure.

Brassard
1-2 cmTuyau d’air
Marque de position de l’artère

2. Une compression de la partie
supérieure du bras, provoquée parun retroussement trop serré de lamanche, peut empêcher l'obtention
de mesures précises.
se trouve
3. Confirmez que l’index
à l’intérieur de la plage d’accord
appropriée.

Français

Veillez à bien fixer
l'attache à tissu.

Index

11

et la plage d’accord approriée

Utilisation de l’appareil
Symboles imprimés sur le brassard
SymbolesFonction/Description

Action recommandée
Réglez la marque ● sur l’artère dubras supérieur ou alignez-la surl'annulaire sur l'intérieur du bras.

Marque de posiotion de l’artère

▲
REF

Index
Numéro du catalogue
Plage d’ajustement appropriée pour lebrassard adulte.
Imprimé sur le brassard adulte.
Utilisez le brassard adulte dePlage supérieure imprimée sur le brassard grande taille au lieu du brassard
adulte et sur le brassard de grande taile. adulte ou du brassard de grande
taile.
Plage appropriée pour le brassard de
grande taille.
Imprimé sur le brassard de grande taille.
Plage d’ajustement appropriée pour lebrassard adulte de grande taille.
Imprimé sur le brassard adulte de grande
taille.
Plage inférieure imprimée sur le brassard
adulte et sur le brassard de grande taille.
Plage inférieure imprimée sur brassard
adulte de grande taille.

Utilisez le brassard adulte au lieu
du brassard adulte de grande taille.

Numéro de lot
Plage d’accord appropriée
Brassard adulte
de grande taille

A

L

Brassardde grande taille

S

W

L

Brassard adulte

S

A

L

Français

12

Comment faire des mesures correctes
Pour une meilleure précision des mesures de la pression sanguine :Asseyez-vous confortablement sur une chaise. Posez le bras sur la table. Ne
croisez pas les jambes. Gardez les pieds au sol et tenez le dos bien droit.
Se relaxer pendant 5 à 10 minutes avant de commencer les mesures.
Lever le bras de sorte que le centre du brassard soit au même niveau que le cœur.
Bouger le moins possible et rester calme pendant les mesures.
Ne pas faire de mesure juste après un effort physique ou un bain. Se
reposer pendant 20 à 30 minutes avant de faire les mesures.
S’efforcer de mesurer la tension à la même heure tous les jours.

Pendant la mesure
Pendant la mesure, il est normal que l'on sente que le brassard est très
serré. (Ne pas s'inquiéter.)

Après la mesure
Après la mesure, appuyez sur le bouton START pour éteindre l’appareil.
Enlevez le brassard et notez votre tension et votre pouls.
Note: Cet appareil est doté de la fonction de mise hors tension automatique,qui éteint un appareil automatiquement dans un délai d’une minuteaprès la mesure.
Attendez au moins trois minutes avant de refaire les mesures sur lamême personne.

Français

13

Les mesures
Avant la mesure, voir «Consignes et observations pour une mesure correcte»
à la page suivante.

Mesure normale

Au niveau du cœur1.

Asseyez-vous confortablement en mettant lebrassard au bras (gauche de préférance).

2.

Appuyez sur le bouton START (mise en marche).
Appuyez
Tous les segments de l’affichage apparaissent.
Puis, zéro (0) s’affiche en clignotant un instant.
Puis, l’affichage change comme indiqué dans la
Tous les segments defigure de droite, lorsque la mesure commence. l’affichage apparaissent
Le brassard commence à gonfler. Il est normal
brièvement.
que l’on sente que le brassard serre fortement lebras. Un indicateur de progression du gonflage
est affiché, dans l'angle gauche de la fenêtre,pendant le gonflage.
Zéro s’affiche
Note: Si vous désirez arrêter le gonflage
quelque que soit le moment, appuyez de Gonflage commence
nouveau sur le bouton START (miseen marche).

3.

Lorsque le brassard est entièrement gonflé, ilcommence à dégonfler automatiquement et le) apparaît poursymbole en forme de cœur (indiquer que la mesure est en cours.
Une fois le pouls détecté, le symbole clignote àchaque pulsation.
Note:

4.

5.

START

Mise souspression

Si la pression correcte n'est pas obtenue,
l'appareil recommence le gonflage du brassard
automatiquement. Pour éviter un nouveau
gonflage, voir la page suivante «Mesure avec
la pression systolique désirée».

Lorsque les mesures sont finies, les lecturesdes pressions systoliques et diastoliques ainsi
que le nombre de pulsations sont affichées.
Le brassard finit de se dégonfler en chassantl'air restant.
Appuyer sur le bouton START pouréteindre l’appareil.

Mesurageen cours

Pression systoliquePression diastoliqueClassification suivant l’OMS

Note:

Attendez au moins trois minutes avant derefaire les mesures sur la même personne.

Pouls
L’air restant est chassé
automatiquement

Français

14

Les mesures
Mesure avec la pression systolique désirée
Le Modèle UA-767S est étudié pour détecter le pouls et pour gonfler lebrassard jusqu’au niveau de la pression systolique automatiquement.
Utilisez cette méthode si un nouveau gonflage se produit à plusieurs reprises, si
les résultats ne s'affichent pas même si la pression diminue de 20 mmHg ou
moins.
1.
2.
3.

4.

Placez le brassard sur le bras au niveau
du cœur (bras gauche de préférence).
Appuyez sur le bouton _START_ .
Lorsque le zéro clignote, appuyez sur lebouton _START_ et maintenez-le enfoncé
jusqu'à ce qu'un nombre de 30 à 40
mmHg supérieur à la pression systoliqueprévue s'affiche.

Au niveau du cœurAppuyez sur le START
boutonRelâcher le bouton
quand la pression
systolique désirée est
obtenueAppuyez et maintenez START
enfoncé le bouton
pour

START
Relâcher le boutoncommencer la mesure, quand le nombre
désiré est indiqué. Ensuite continuez àmesurer votre tension artérielle commedécrit à la page précédente.

Se référer à la page précédente
pour la mesure.

Consignes et observations pour une mesure correcte

Asseyez-vous confortablement. Posez le bras sur une table, la paume dirigéevers le haut et le brassard au même niveau que le cœur.
Détendez-vous pendant cinq à dix minutes avant de commencer la mesure. Si
vous êtes énervé ou au contraire déprimé, cela se reflètera sur votre tension demême que sur votre pouls.
La tension de tout individu varie constamment selon ce qu’il fait et ce qu’ilmange. D’autre part, certaines boissons ont un effet très fort et très rapide surla tension artérielle.
Cet appareil se réfère aux pulsations cardiaques pour mesurer la tension et lepouls. Par conséquent, si vos pulsations sont faibles ou irrégulières, l’appareil
pourrait avoir des difficultés à mesurer votre tension artérielle.

Au cas où l’appareil détecte une condition anormale, il s’arrête de mesurer etaffiche un symbole d’erreur. Voir la page 7 pour la description des symboles.
Français

15

Cet appareil est conçu pour être utilisé par des adultes. Avant de l’utiliser pour
mesurer la tension d’un enfant, veuillez en discuter avec votre médecin. Ne
laissez jamais un enfant prendre lui-même sa tension sans surveillance.
La performance du tensiomètre automatique peut être affectée par unetempérature, une humidité ou une altitude excessives.

Rappel des données mémoriséesNote: Cet appareil mémorise les 60 dernières mesures.
1. Si rien ne s'affiche, appuyez sur lebouton _START_ et maintenez-leenfoncé pour rappeler les données
enregistrées.

Appuyez et maintenez START
enfoncé le bouton
Pression systolique
moyennePression diastolique
moyennePouls moyen
Moyenne de toutes
les données

2. Relâchez le bouton lorsque les données
moyennes s'affichent.
3. Le numéro des données et les données
enregistrées s'affichentautomatiquement dans l'ordre, à partirde la dernière mesure.

PressionsystoliquePression
diastolique
Pouls

4. L'écran s'éteint automatiquement une fois
toutes les données affichées.
Remarque : Si vous appuyez sur le bouton_START_ lorsque vous rappelez
des données, l'appareil s'éteint.

Données les plus récentes

PressionsystoliquePression
diastolique
PoulsDernières données (les plus anciennes)

Français16

Qu’est ce que l’indicateur de RCI/FA?
Lorsque le moniteur détecte un rythme irrégulier pendant les mesures,l’indicateur de RCI/FA apparaît à l’écran avec les valeurs mesurées.
Note: Nous vous recommandons de contacter votre médecin si vous voyez
fréquemment cet indicateur de RCI/FA :.

Qu’est-ce que la FA?
Le cœur se contracte en raison des signaux électriques se produisant dans lecœur et envoie le sang dans tout le corps. La fibrillation auriculaire (FA) se
produit lorsque le signal électrique dans l’oreillette est troublé et provoque desperturbations dans l’intervalle des pulsations. La FA peut causer la stagnation
du sang dans le cœur, ce qui peut facilement créer des caillots sanguins, cause
d’AVC et de crise cardiaque.

％RCI/FA%RCI /FA exprime la fréquence de RCI/FA détecté.
Le RCI/la FA peut détecter non seulement les parasites tels que des mouvements
physiques, mais également un rythme cardiaque irrégulier. Par conséquent, nous
vous recommandons de contacter votre médecin si le niveau du %RCI/FA est élevé.
Nombre de RCI/FA
détectés en mémoire
x 100 [%]
%RCI/FA =Nombre total
Affichage du %RCI/FA: le %RCI/FA apparaît lors de l’affichage des valeursmoyennes. (Reportez-vous à « 2. Rappel des données enmémoire » dans « Mode opératoire »)Le %RCI/FA ne s’affiche pas si le nombre de mémoire est
de six ou moins.
Affichage de la valeur moyenne

Niveau 0%RCI/FA=0

Niveau 1%RCI/FA=1 - 9

Niveau 2
Niveau 3%RCI/FA=10 - 24 %RCI/FA=25 - 100

Non affiché

Français

17

Indicateur de progression du gonflage
L’indicateur montre la progression de la pression pendant la mesure.
Mise sous pression

Dégagement d’air

Gonflage en cours

Mesure en cours
Gonflement complet

Indicateur de tension selon la classification OMS
Chaque segment d’indicateur de progression
de la pression correspond à la classification dela pression sanguine selon l’OMS décrite à lapage suivante.
Indicateur de la classification selon l’OMS
Hypertention sévère
Hypertention modérée
Hypertention faible
Haute-Normale
NormaleOptimale

Exemple:Hypertention modérée

Hypertention faible: L’indicateur affiche un segment, basé surles dernières données, correspondant à laclassification suivant l’OMS.
Haute-Normale

Pression sanguine
Qu'est-ce que la pression sanguine?
La pression sanguine est la force avec laquelle le sang vient frapper contre lesparois des artères. La pression systolique se produit lorsque le cœur secontracte. La pression diastolique se produit lorsque le cœur se détend.
La pression sanguine est mesurée en millimètre de mercure (mmHg). La
pression sanguine naturelle d’un individu est représentée par la pression
fondamentale qui est mesurée avant toute activité, le matin au lever du lit, etavant de manger ou de boire.
Français18

Qu'est-ce que l'hypertension et comment la contrôler?
L’hypertension est une tension artérielle (ou plus exactement une «pression
sanguine») anormalement élevée. Non surveillée, l’hypertension peut provoquer
de nombreux problèmes de santé, notamment une attaque d’apoplexie ou une
crise cardiaque.
L'hypertension peut être contrôlée en changeant son style de vie, en évitant
tout stress et avec des médicaments pris sous surveillance médicale.
Pour empêcher l’hypertension ou la garder sous contrôle:
Ne pas fumer.
Faire régulièrement des exercices
Réduire sa consommation de sel etphysiques.
Effectuer régulièrement desde graisse.
Maintenir un poids adéquat.
visites médicales.

Comment mesurer la pression sanguine chez soi?
Lorsque la tension est mesurée à l’hôpital ou dans le cabinet du médecin, on neparvient pas toujours à se décontracter, ce qui peut avoir pour effet d’augmenterla tension de 25 à 30 mmHg par rapport à la tension mesurée chez soi. Le fait demesurer sa tension tranquillement chez soi réduit les influences extérieures surla valeur de la tension, complète les mesures effectuées par le médecin etpermet d’établir un relevé plus précis et plus complet de sa tension artérielle.

Classification de la tension selon l’OMS
L’évaluation de l’hypertension,
indépendamment du critère âge, a étéétablie par l’Organisation Mondiale de laSanté (OMS) selon le tableau ci-joint.

Variations de la pression sanguine
La tension artérielle (ou pression
sanguine) d’un individu varie fortement
selon le moment de la journée et selon lasaison. Ainsi, elle peut varier de 30 à 50
mmHg dans une même journée selon un ensemble de conditions. Chez lespersonnes hypertendues, ces variations sont souvent encore plus prononcées.
Normalement, la pression augmente pendant le travail ou le jeu et tombe à sonplus bas niveau pendant le sommeil. Par conséquent ne pas trop s'inquiéter
des résultats d'une seule mesure.
Mesurez votre tension tous les jours à lamême heure (en suivant la procédure
décrite dans ce manuel) pour connaître
votre pression sanguine normale. Ces
mesures quotidiennes vous permettrontd’établir un relevé plus complet de votre
tension.
Veillez à bien noter la date et l'heure
lorsque vous enregistrez votre pression.
Consultez votre médecin pour interpréter
les données recueillies.
Français

19

Résolution des problèmes
Problème

Cause probable

Solution recommandéeLes piles sont usées.
Rien n'apparaît
sur l'affichage,même lorsque l'on
Les bornes des piles ne sont
met l'appareil en
pas dans le bon sens.
marche.

Remplacer les piles usées pardes neuves.
Mettre les piles dans le bonsens en faisant correspondre laborne positive et négative
comme indiqué sur le brassard.

La tension des piles est trop
faible.
Remplacer les piles usées par
Le brassard ne se
Le symbole piles faibles
gonfle pas.
des neuves.
apparaît. Si les piles sont usées,ce symbole n’apparaît pas.
Serrer correctement le
Le brassard est mal serré.
brassard.
Vous avez bougé votre bras ou Ne pas bouger et rester calme
autre pendant la mesure.
pendant la mesure.
L’appareil ne faitS'asseoir confortablement et ne
pas les mesures.
pas bouger. Posez le bras sur
Les lectures sont Le brassard est mal placé.
une table, la paume dirigée vers
trop hautes ou
le haut et le brassard au mêmetrop basses.
niveau que le cœur.
Si votre pouls est faible ou
irrégulier, l'appareil aura desdifficultés à déterminer votre
pression sanguine.
La valeur est différente de celle Voir le paragraphe "Comment
mesurée dans une clinique ou mesurer la pression sanguinechez soi?".
chez le médecin.
AutresRetirer les piles. Le remettre
correctement en place etessayer à nouveau de mesurer.
Note: Si les solutions décrites ci-dessus ne suffisent pas pour résoudre les problèmes,contactez votre revendeur. N’essayez pas d’ouvrir ou réparer vous-même cetappareil car toute tentative invaliderait votre garantie

Français

20

Entretien
Ne pas ouvrir l'appareil car il comporte des composants électriques sensibles et
une unité d'air complexe qui pourraient être endommagés. Si vous n'arrivez pasà situer un problème avec le tableau de recherche de problèmes, demandez auservice après vente de votre revendeur ou du groupe de service A & D. Le
groupe de service A & D vous fournira des informations techniques sur lesfournisseurs agréés, les pièces et les unités de rechange.
Cet appareil a été conçu et fabriqué pour durer. Il est généralement
recommandé de le faire vérifier tous les 2 ans pour s’assurer de son
fonctionnement correct et de sa précision. Veuillez contacter votre revendeur
agréé A&D pour l’entretien.

Fiche techniqueType
Méthode de mesure
Plage de mesure

Précision de mesure
Alimentation électriqueNombre de mesures

Classification

Test clinique

EMD
Mémoire de données
Condition de fonctionnement
Condition de stockage / transport

UA-767S
Mesure oscillométriquePression:de 0 à 299 mmHg
Pression systolique : de 60 à 279 mmHg
Pression diastolique : de 40 à 200 mmHg
Pouls:de 40 à 180 pulsations / minutePression:±3 mmHg
Pouls:±5%
4 x 1.5V piles (R6P, LR6 ou AA) ou
adaptateur secteur (TB-233C) (Non fournis)
Env. 700 fois LR6 (piles alcalines)
Env. 200 fois R6P (piles au manganèse)
Avec une valeur de pression de 180 mmHg,
température ambiante de 23 °C.
Équipement a alimentation interne ME (fournie parles piles) /Class II (fournie par l'adaptateur)Mode de fonctionnement en continu
Conformément à ISO81060-2：2013
Dans l’étude de validation clinique, K5 a été utilisé
sur 85 sujets pour la détermination de la tension
artérielle diastolique.
IEC 60601-1-2: 2014
Dernières 60 mesuresde +10 à + 40 °C / hum. rel. de 15 à 85 %/de 800 à 1060 hPa
de -20 à + 60 °C / hum. rel. de 10 à 95 % /de 700 à 1060 hPa
Français21

Dimensions
Poids
Protection
Partie appliquée
Durée de vie utile

environ 140 [l] x 60 [h] x 105 [p] mmenviron 245 g sans les piles
Appareil: IP20
Brassard Type BF
Appareil: 5 ans (six utilisations par jour)Brassard : 2 ans (six utilisations par jour)
Adapteur secteur : 5 ans (lorsqu'utilisé six fois par jour)

Adaptateur secteur accessoire
L'adaptateur permet de brancher l'appareil sur une source d'alimentation
domestique. Veuillez contacter votre distributeur A&D local pour l'acheter.
L'adaptateur secteur doit être périodiquement inspecté ou remplacé.
L’adaptateur secteur doit être inspecté ou remplacé
périodiquement.

Adaptateur secteur
TB-233C

Symboles imprimés sur l’adaptateur secteur
SymbolesFonction / Signification
Utilisation à intérieur uniquement
Appareil de classe II
Fusible thermique
FusibleLabel de conformité de l’appareil à ladirective CE
Label de conformité de l’appareil à lacertification EAC
Polarité de la fiche de l’adaptateur secteur
Accessoires vendus séparément
Brassard

Numéro du catalogue
CUF-F-LA
CUF-I
CUF-F-A

Taille du brassard
Brassard adulte degrande tailleBrassard de grande
tailleBrassard adulte

Taille du bras
31 cm à 45 cm
22 cm à 42 cm
22 cm à 32 cm

Taille du bras：La circonférence du biceps
Adaptateur secteur

Note:

Numéro du catalogue
TB-233C

Fiche
Type CCes caractéristiques techniques sont sujettes à changement sans préavis.
Le classement IP indique les degrés de protection fournis par les coffrets deprotection conformément à la norme CEI 60529. Cet appareil est protégé contre lescorps étrangers solides de 12 mm de diamètre et plus, tel qu’un doigt. Cet appareiln’est pas protégé contre l’eau.

Français22ÍNDICE
Estimados clientes ................................................................ 2
Observaciones Preliminares .................................................. 2
Precauciones ........................................................................ 2
Partes de identificación de componentes.................................. 5
Símbolos .............................................................................. 6
Modo de operación ............................................................... 8
Uso del monitor .................................................................... 9
Instalación/Cambio de las pilas........................................ 9Conexión de la manguera de aire ................................... 10
Conexión del adaptador de CA ....................................... 10
Selección del brazalete correcto ..................................... 11
Cómo colocar el brazalete .............................................. 11
Para efectuar mediciones correctas ................................ 13
Medición ........................................................................ 13
Después de la medición ................................................. 13
Mediciones ......................................................................... 14
Medición normal ............................................................ 14
Medición con la presión sistólica deseada ...................... 15
Notas para mediciones correctas..................................... 15
Revisión de los datos de la memoria ................................... 16¿ Qué es el indicador de LI/FibA? ........................................ 17¿A qué hace referencia FibA? ............................................... 17％LI/FibA ............................................................................ 17
Indicador de barra de presión ............................................. 18
Indicador de clasificación de la OMS ................................... 18
Aspectos sobre la tensión arterial ....................................... 18¿Qué es la tensión arterial?............................................. 18¿Qué es la hipertensión y cómo controlarla? ................... 19¿Por qué debe medirse la tensión arterial en la casa? ...... 19
Clasificación de tensión arterial según la OMS ................ 19
Variaciones de tensión arterial ....................................... 19
Localización y corrección de fallos ...................................... 20
Mantenimiento ................................................................... 21
Datos técnicos .................................................................... 21

Español

1

Estimados clientes
FELICIDADES. Usted ha comprado un monitor de presiones sanguíneas A&D,
el aparato tecnológico más avanzado y más fácil de usar actualmente en el
mercado.
Se recomienda leer este manual de instrucciones detenidamente antes deusarlo por primera vez.

Observaciones Preliminares
Este aparato se ajusta a la Directiva Europea 93/42 EEC para productosde conformidad. (0123:
médicos. Esto se evidencia con la marca
Número de referencia a los quienes notificados)
Este aparato está diseñado para usarse en adultos, los niños y los recién nacidos
no pueden usarlo.
Ámbito de uso. Este aparato es para que pueda usarlo usted mismo en un entorno
doméstico.
Este aparato se ha diseñado para medir la presión sanguínea y la frecuenciadel pulso de las personas para el diagnóstico.

Precauciones
En la construcción de este aparato, se han usado componentes de precisión.
Deberán evitarse extremos de temperatura, humedad, rayos solares directos,
golpes o polvo.
Limpie el aparato y el brazalete con un paño seco, suave o un paño
humedecido con agua y un detergente neutro. No use nunca alcohol, gasolina,
diluyente u otros productos químicos para limpiar el aparato o el brazalete.
Evite doblar el brazalete apretado o guardar el tubo retorcido duranteperíodos prolongados, dado que ello puede acortar la vida útil de loscomponentes.
Tenga cuidado para evitar la estrangulación accidental de niños o bebés con
el tubo o el cable.
No retuerza la manguera de aire durante la medición. Si lo hace, pueden
producirse lesiones debido a la presión continua del brazalete.
El aparato y el brazalete no son resistentes al agua. Evite que la lluvia, latranspiración y el agua mojen el aparato y el brazalete.
Las mediciones pueden verse afectadas si el aparato se usa cerca detelevisores, hornos microondas, teléfonos móviles, rayos X u otros dispositivos
con campos eléctricos fuertes.
Los aparatos de comunicación inalámbrica, como los dispositivos de red
doméstica, teléfonos móviles, teléfonos inalámbricos y sus centrales ywalkie-talkies, pueden afectar al monitor de presiones sanguíneas. Por lotanto, es necesario mantener una distancia de 30 cm entre el monitor ydichos aparatos.
Español2

Cuando se reutiliza el aparato, confirme que el aparato está limpio.
El aparato, sus partes y pilas usados no deben considerarse como
desperdicios domésticos ordinarios y deberán eliminarse de acuerdo con losreglamentos locales aplicables.
Cuando se usa un adaptador CA, asegúrese de que el adaptador CA sepuede retirar rápidamente del enchufe eléctrico cuando sea necesario.
No modifique el aparato. Puede provocar accidentes o daños al aparato.
Para medir la presión sanguínea, se debe apretar el brazo con el brazalete lo
suficiente para detener el flujo de sangre a través de la arteria temporalmente.
Esto puede provocar dolor, hormigueo o una marca roja temporal en el brazo.
Este estado aparecerá especialmente cuando la medición se repite
sucesivamente. Cualquier dolor, hormigueo o marcas rojas desaparecerán
con el tiempo.
Si la presión sanguínea se mide con demasiada frecuencia, pueden
producirse daños debido a la interferencia con el flujo sanguíneo. Si utiliza elaparato de forma repetida, compruebe que su uso no provoca una alteración
prolongada de la circulación sanguínea.
Si le han practicado una mastectomía, consulte con su médico antes deutilizar el aparato.
No deje que los niños utilicen el dispositivo sin la supervisión de un adulto, ni
utilice el dispositivo en un lugar dentro del alcance de los niños. Podría causar
accidentes o daños.
El aparato contiene piezas pequeñas que pueden provocar un atragantamiento
si los niños las tragan accidentalmente.
Desconecte el adaptador de CA cuando no se use durante la medición.
El uso de accesorios no indicados en este manual puede afectar a la seguridad.
Si se produce un cortocircuito en la pila, puede calentarse y posiblementecausar quemaduras.
Permita que el aparato se ajuste al entorno circundante antes de usarlo(alrededor de una hora).
No se han realizado estudios clínicos en bebés recién nacidos o mujeres
embarazadas. No lo use en niños recién nacidos o mujeres embarazadas.
No toque las pilas, la clavija de CC y al paciente simultáneamente. Podría
ocasionar una descarga eléctrica.
No lo hinche sin colocar el brazalete en la parte superior del brazo.

Contraindicaciones
A continuación se indican precauciones para el uso correcto del aparato.
No aplique el brazalete a un brazo con otro aparato eléctrico médico acoplado.
Es posible que el aparato no funcione correctamente.
Las personas con un grave déficit circulatorio en el brazo deben consultar aun médico antes de utilizar el aparato, para evitar problemas médicos.
No autodiagnostique los resultados de medición ni inicie el tratamiento usted
mismo. Consulte siempre a su médico para la evaluación de los resultados yel tratamiento.
Español3No aplique el brazalete a un brazo con una herida sin curar.
No aplique el brazalete a un brazo que está recibiendo un goteo intravenoso
o transfusión sanguínea. Puede provocar lesiones o accidentes.
No utilice el aparato en donde haya gases inflamables como, por ejemplo,
gases anestésicos. Puede provocar una explosión.
No utilice el aparato en entornos con alta concentración de oxígeno, tal comolas cámaras de oxígeno de alta presión o tienda de oxígeno. Puede provocar
un incendio o explosión.

Español

4Partes de identificación de componentes
Pantalla

Tapón delconector de aire

Enchufeadaptadorde CAJack de CC

Zócalo de aire

Botón START
Manguerade aireMarca de posición
de la arteria

Rango de colocación

Marca índice
BrazaleteCompartimiento de pilas

Cubierta de pilas

Pilas de 1,5 V(R6P, LR6 o AA)

Indicación de la pantalla

Promedio

Memoria

Símbolo LI/FibA

Presión
sistólica%LI/FibA

Símbolo de error de ajuste del brazalete

Presión
diastólica

Símbolo de error por movimiento
Marca de corazón

Frecuencia del pulso

Indicador de clasificación de OMS
e indicador de la barra de presión
Indicador de pilas
Español5

Símbolos
Símbolos impresos en la carcasa del aparato
SímbolosFunción / Significado
Aparato en “standby” y botón de encendido

SYS

Presión sanguínea sistólica indicada en mmHg.

DIA

Presión sanguínea diastólica indicada en mmHg.

PUL

Frecuencia del pulso por minuto
Guía para instalación de pilas
Corriente continua
Tipo: BF; El aparato, el brazalete y tubo están diseñados para
proporcionar protección especial contra electrochoques.
Etiqueta de aparatos médicos según directiva de CE
Representante UE
FabricanteFecha de fabricación

IP

Símbolo de protección internacional
Etiqueta de WEEE

SN

Número de serie
Consulte el manual/folleto de instrucciones
Polaridad del jack de CC
Manténgase seco

Símbolos que aparecen en la pantalla
SímbolosFunción / Significado

Acción recomendada

Aparece cuando la medición está en
curso. Este indicador parpadea cuando La medición está en curso.
se detecta el pulso.
Símbolo LI/FibA
Este indicador aparece cuando se
detectan latidos irregulares del corazón.
Puede iluminarse cuando se detecta
una vibración muy ligera, como un
escalofrío o un temblor.
La lectura puede ofrecer
Símbolo que aparece cuando se
un valor incorrecto.
detecta un movimiento del cuerpo o el Vuelva a medir otra vez.
brazo.
Quédese muy tranquilo
durante la medición.
Español

6

Símbolos
SímbolosFunción / Significado

Acción recomendada

Símbolo que aparece durante la
medición cuando el brazalete no estábien ajustado.

La lectura puede ofrecer
un valor incorrecto.
Coloque el brazaletecorrectamente, y vuelva amedir otra vez.

Índice detectado de LI/FibA en la memoria
Número detectado de
LI/FibA en la memoria
x 100 [%]
%LI/FibA =Número total
Las mediciones previas almacenadas
en la memoria.
Datos medios
PILAS CARGADAS
Indicador de la potencia de las pilas
durante la medición.
PILAS GASTADAS
Reemplace todas las pilas
Las pilas son débiles cuando parpadea con unas nuevas, cuando
esta marca.
destelle esta marca.
Vuelva a medir otra vez.
Presión sanguínea inestable debido al
Quédese muy tranquilo
movimiento durante la medición.
durante la medición.
Los valores sistólico y diastólico seencuentran dentro de 10 mmHg.
El valor de presión no ha aumentado
Coloque el brazalete
durante la inflación.
correctamente, y vuelva a
El brazalete no está colocado
medir otra vez.
correctamente.
ERROR DE VISUALIZACIÓN DEL PULSOEl impulso no se ha detectado
correctamente.

Error interno del monitor de presiones
sanguíneas

Español7

Extraiga las pilas, presione elbotón START y acontinuación vuelva a colocar
las pilas. Si el error sigue
apareciendo, póngase en
contacto con su distribuidor.

Modo de operación
1. Medición normal con los datos almacenados
Presione el botón START . Se mide la presión sanguínea y sus datos sealmacenan en el aparato. El aparato puede almacenar los últimos 60 datos
automáticamente en la memoria.

2. Revisión de los datos de la memoria En "standby",
Cuando no se visualice nada, mantengapresionado el botón START .

mantenga presionado
el START botón

Suelte el botón cuando se muestren los datos
promedio.

El número de dato y los datos guardados semuestran automáticamente en el orden desdela última medición.

Para más detalles acerca de la revisión de datos,
consulte “Revisión de los datos de la memoria”.

3. Borrado de datos los datos almacenados en la memoria
Cuando apague el aparato, pulse y mantenga pulsado elbotón _START_ hasta que se muestre "".
Seleccione "" para borrar los datos.
.
El dato se borra cuando parpadea la marca
El aparato se apaga automáticamente.

START

4. Medición con la presión sistólica deseada
Véase página 15 para la medición con la presión sistólica deseada.

Español8

Uso del monitor
Paso 1

Instalación/Cambio de las pilas

Paso 3

1. Retire la cubierta de pilas.
2. Retire las pilas usadas e inserte lasnuevas en el compartimiento tal como
se describe, respetando las polaridades(+ y -) de las pilas.
Utilice solo las pilas R6P, LR6 o AA.

Paso 1

Paso 2

3. Cierre la cubierta de pilas.

PRECAUCIÓN
Coloque las pilas en el compartimiento de las pilas, tal como se
muestra en la figura. De lo contrario, el aparato no funcionará.
Cuando la marca(marca de PILAS GASTADA) parpadea en la pantalla, reemplace
todas las pilas por las nuevas. No mezcle pilas viejas con las nuevas, dado queesto puede acortar la vida útil de las mismas o causar un mal malfuncionamientodel aparato.
Sustituya las pilas al menos dos segundos después de que se apague el aparato.
(marca de PILAS GASTADA) aparece incluso después de sustituir las
Si la marcapilas, realice una medición de la presión sanguínea. El aparato podría entonces
reconocer las pilas nuevas.
La marca(marca de PILAS GASTADA) no aparece cuando las pilas están totalmenteagotadas.
La vida útil de las pilas varía con la temperatura ambiente y puede acortarse atemperaturas bajas. Por lo general, cuatro pilas LR6 nuevas duran
aproximadamente un año si se usan para realizar mediciones dos veces al día.
Utilice solamente las pilas especificadas. Las pilas suministradas con el aparato sonpara probar el funcionamiento del monitor y pueden tener una vida útil más corta.
Retire las pilas cuando no vaya a usar el aparato durante un período prolongado.
Las pilas pueden tener fugas y causar un malfuncionamiento.

Español9

Uso del monitor
Conexión de la manguera de aire

Zócalo de aire

Inserte firmemente el tapón delconector de aire en el zócalo de aire.

Tapón del conector de aire

Conexión del adaptador de CA
Inserte el enchufe del adaptador de CA
en el jack de CC.
A continuación, inserte el adaptador de
CA en una toma de corriente.

Jack de CC

Enchufe adaptador de CA
Utilice el adaptador de CA especificado.
(Véase página 22.)
Cuando desconecte el adaptador de CA de la toma de corriente, agarre elarmazón del adaptador de CA y tire de él para extraerlo de la toma decorriente.
Cuando desconecte el enchufe del adaptador de CA del monitor de presiones
sanguíneas, agarre el enchufe del adaptador de CA y tire de él para extraerlodel monitor.

Español

10Uso del monitor
Selección del brazalete correcto
La selección de un brazalete de tamaño correcto es importante para la lectura
correcta. Si el brazalete no es de tamaño correcto, la lectura puede indicar un
valor incorrecto de la presión sanguínea.
El tamaño del brazo está impreso en cada brazalete.
El índicey el gama adecuada de colocación del brazalete le indicarán, si elbrazalete está colocado correctamente. (Refiérase a "Símbolos impresos en el
brazalete" en la página siguiente.)
Si el índice
se encuentra fuera de la gama, póngase en contacto con el
distribuidor local para adquirir el brazalete de reemplazo.
El brazalete es un producto consumible. Si se deteriora, adquiera uno nuevo.
Tamaño del brazo Tamaño del brazalete recomendado
Nº de catálogo
31 cm a 45 cm
Brazalete grande de adulto
CUF-F-LA
22 cm a 42 cm
Brazalete de amplio rango
CUF-I
22 cm a 32 cm
Brazalete de adulto
CUF-F-A
Tamaño del brazo: La circunferencia en el bíceps.
Nota:
El modelo UA-767S no se ha diseñado para usar un
brazalete pequeño de adulto.

Cómo colocar el brazalete

Evite que la manga quede demasiado
apretada al remangarse.

1. Coloque el brazalete alrededor del brazo,
unos 1-2 cm por encima de la parteinterior del codo, tal como se muestra en
la figura. Coloque el brazalete encontacto directo con la piel, dado que laropa puede causar un pulso débil yprovocar error de medición.

Brazalet
1-2 cmManguera de aire
Marca deposición de la arteria

2. Asimismo, la constricción del brazo,
causada al remangarse la manga
apretada, no permitirá lecturas correctas.

Abroche bien elcierre de velcro.

se encuentre
3. Confirme que el índicedentro de la gama de buen estado.
ÍndiceEspañol11y la rango de colocación

Uso del monitor
Símbolos impresos en el brazaleteSímbolosFunción/Significado●

Marca de posición de la arteria▲
REFÍndiceAcción recomendada
Ajuste la marca ● en la arteria delbrazo o de forma alineada con el dedo
anular en la parte interior del brazo.

Número de catálogo

Rango de colocación adecuado parael brazalete de adulto.
Está impreso en el brazalete de adulto.
Por encima del rango impreso en el Utilice el brazalete grande de adulto
brazalete de adulto y el brazalete de en lugar del brazalete de adulto oamplio rango.
brazalete de amplio rango.
Rango de colocación adecuado para el
brazalete de amplio rango.
Está impreso en el brazalete de amplio
rango.
Rango de colocación adecuado parael brazalete grande de adulto.
Está impreso en el brazalete grande deadulto.
Por debajo del rango impreso en el
brazalete de adulto y el brazalete deamplio rango.
Utilice el brazalete de adulto en
Por debajo del rango impreso en ellugar del brazalete de adulto
brazalete de adulto grande.
grande.
Número de loteRango de colocación
Brazalete grande de adulto

A

L

Brazalete de amplio rango

S

W

L

Brazalete de adulto

S

A

L

Español

12

Para efectuar mediciones correctas
Tenga en cuenta lo siguiente para una óptima precisión en la medición de la tensión arterial:
Siéntese cómodamente en una silla. Apoye su brazo sobre la mesa. No cruce
las piernas. Mantenga los pies en el suelo y enderece la espalda.
Relájese durante unos 5 a 10 minutos.
Coloque el centro del brazalete al mismo nivel que su corazón
Permanezca quieto y en silencio durante la toma de la presión.
No efectúe la medición inmediatamente después de un ejercicio físico o baño.
Descanse unos 20 ó 30 minutos antes de la medición.
Trate de medir su tensión arterial a la misma hora todos los días.

Medición
Durante la medición, es normal que se sienta el brazalete muy apretado (no se
alarme por esto).

Después de la medición
Después de la medición, presione el botón _START_ para apagar el aparato.
Retire el brazalete y registre sus datos de medición.
Nota: El aparato cuenta con la función de desconexión automática de laalimentación, que desconecta la alimentación automáticamente un minuto
después de la medición.
Deje que transcurran por lo menos tres minutos entre mediciones de lamisma persona.

Español

13

Mediciones
Antes de la medición, lea “Notas para mediciones correctas” en la página
siguiente.

Medición normal
1.

Coloque el brazalete en su brazo (brazo
izquierdo preferiblemente).
Siéntese tranquilamente durante la medición.

2.

Presione el botón _START_.
Aparecen brevemente todas las secciones dela pantalla. A continuación, parece cero (0)parpadeando brevemente. Luego la indicaciónde la pantalla cambia como se muestra en la
figura derecha, al comenzar la medición. El
brazalete comienza a inflarse. Es normal que
el brazalete apriete fuertemente. Se visualiza
el indicador de barra de presión durante la
inflación en el borde izquierdo de la pantalla.
Nota:

3.

5.

Presione

START

Aparecen brevemente
todas las secciones
de la pantalla.

Aparece ceroComienza la inflación

Al término de la inflación del brazalete, ladeflación comienza automáticamente y lamarca(marca de corazón) parpadea
indicando que la medición está en curso. Unavez detectado el pulso, esta marca parpadeará
coincidiendo con el latido del corazón.
Nota:

4.

Si desea detener el inflado en cualquier
momento, presione el botón _START_
nuevamente.

A nivel del corazón

Presionamiento

Si no se obtiene una tensión apropiada, el
aparato empieza nuevamente el inflado
automáticamente.
Para evitar la reinflación, lea el apartado“Medición con la presión sistólica
deseada” en la página siguiente.

Al término de la medición, se visualizan laslecturas de las tensiones sistólica y diastólica
y frecuencia del pulso.
El brazalete expulsa el aire remanente y se
esinfla completamente.
Presione el botón _START_ para apagar elaparato.

Medición
progresiva

Presión sistólica
Presión diastólica
Clasificación según la OMS
Frecuencia del pulso

Nota: Deje que transcurran por lo menos tres minutos
entre mediciones de la misma persona.
Español

14

Evacuación automática
del aire remanente.

Mediciones
Medición con la presión sistólica deseada
El modelo UA-767S está diseñado para detectar el pulso, inflando
automáticamente el brazalete hasta el nivel de presión sistólica.
Utilice este método cuando la represurización se produzca de forma repetida, ocuando los resultados no se muestren incluso aunque la presión disminuya a
20 mmHg o menos.

1.

Coloque el brazalete en el brazo en el nivel
del corazón (preferiblemente el brazo
izquierdo).

Al nivel delcorazón
Pulse el botón START

2.

Pulse el botón _START_ .

3.

Mientras el cero parpadea, pulse ymantenga pulsado el botón _START_
hasta que aparezca un número
alrededor de 30 a 40 mmHg superior a laMantenga presionado START
presión sistólica esperada.

Suelte el botón cuando
se visualice la presión
sistólica deseada.

el botón

4.

Suelte el botón _START_ al alcanzar elvalor deseado, ya que la medición
comenzará. Luego continúe la medición
de su presión sanguínea como se
describe en la página anterior.

Para la medición, refiérase
a la página anterior.

Notas para mediciones correctas

Siéntese cómodamente. Coloque el brazo sobre una mesa con la palma hacia
arriba y el brazalete al mismo nivel que el corazón.
Relájese por unos cinco o diez minutos antes de iniciar la medición. Si seencuentra agitado o deprimido debido a un estrés emocional, la medición reflejará
este estrés con una lectura mayor (o menor) que la de su tensión arterial normal yla lectura del pulso generalmente será más rápida que la normal.
La tensión arterial de una persona varía constantemente, dependiendo de lo
que está haciendo y lo que ha comido. Lo que tome podrá tener un efecto muy
fuerte y rápido sobre su tensión arterial.
Las mediciones de este aparato se basan en el latido del corazón. Si usted
tiene un latido muy débil o irregular, este aparato puede tener dificultad en
determinar su tensión arterial.
En caso de que el aparato detecte una condición anormal, el mismo detendrá
la medición e indicará un símbolo de error en la pantalla. Vea la página 7 parala descripción de los símbolos.
Español

15

Este aparato está diseñado para su uso en adultos. Consulte con su médico
antes de usar este aparato en un niño. Un niño no debe usar este aparato sin la
atención de un adulto.
El rendimiento del tensiómetro automático puede verse afectado por temperatura
y humedad excesivas o por la altitud.

Revisión de los datos de la memoria
Nota:
1.

Este aparato almacena en la memoria las últimas 60 mediciones.

Si no se muestra nada, pulse ymantenga pulsado el botón _START_
para recuperar los datos guardados.

Mantengapresionado STARTel botónPresión sistólica media
Presión diastólica media
Frecuencia media del pulso

2.

Suelte el botón cuando se muestren losdatos promedio.

3.

El número de dato y los datos
guardados se muestranautomáticamente en el orden desde laúltima medición.

Promedio detodos los datos

Sistólica

4.

Diastólica

La pantalla se apagará
automáticamente después de habermostrado todos los datos.

Frecuenciadel pulso
Datos más recientes

Nota: Si pulsa el botón _START_ cuando
recupera los datos, el aparato se
apagará.

Sistólica
Diastólica
Frecuenciadel pulsoÚltimo dato (más antiguo)

Español16¿ Qué es el indicador de LI/FibA?
Si el monitor detecta un ritmo irregular durante las mediciones, aparecerá elindicador LI/FibA en la pantalla con los valores de medición.
Nota: Le recomendamos que consulte a su médico si ve el símbolo
LI/FibA con
frecuencia.

¿A qué hace referencia FibA?
El corazón se contrae debido a corrientes eléctricas que surgen de él y envía
sangre por todo el cuerpo. La fibrilación auricular (FibA) se produce cuando las
corrientes eléctricas del atrio se confunden y ocasionan alteraciones en losintervalos del pulso. La FibA puede ocasionar estancamientos de sangre en elcorazón, lo que podría producir fácilmente coágulos de sangre, una de lascausas de las apoplejías y los infartos de corazón.

％LI/FibA%LI/FibA aparece como índice del LI/FibA detectado.
LI/FibA puede detectar no solamente interferencias tales como el movimiento
físico, sino también los latidos irregulares del corazón. Por tanto, lerecomendamos consultar con su médico si el nivel de %LI/FibA es alto.
Número detectado de
LI/FibA en la memoria
x 100 [%]
%LI/FibA =Número total
Visualización de %LI/FibA: %LI/FibA se visualiza al mostrar los valores medios.
(Refiérase a “2. Revisión de los datos de lamemoria” en “Modo de operación”.)%LI/FibA no se visualiza si el número almacenado
en la memoria es igual o inferior a seis.
Visualización de valores medios

Nivel 0%LI/FibA = 0

Nivel 1
%LI/FibA = 1 - 9

Nivel 2
Nivel 3
%LI/FibA = 10 - 24 %LI/FibA = 25 - 100

No se visualiza

Español

17

Indicador de barra de presión
El indicador monitoriza el progreso de la presión durante la medición.
Presurizando

Liberando aire

Inflación en progreso

Medición en progreso
Se ha completado la inflación

Indicador de clasificación de la OMS
Cada uno de los seis segmentos del indicadorde barra corresponde a la clasificación depresión sanguínea de la OMS que se describe
en la siguiente página.

Indicador de clasificación de la OMS
Hipertensión severa
Hipertensión moderada
Hipertensión leve
Normal alta
NormalOptimaEjemplo:

Hipertensión moderada

Hipertensión leve

: El indicador visualiza un segmento basado
en el dato actual, correspondiente a laclasificación de la OMS.

Normal alta

Aspectos sobre la tensión arterial¿Qué es la tensión arterial?
La tensión arterial es la fuerza que ejerce la sangre contra las paredes de lasarterias. La tensión sistólica ocurre cuando el corazón se contrae, y la tensión
diastólica ocurre cuando el corazón se expande.
La presión sanguínea se mide en milímetros de mercurio (mmHg). La presión
arterial de una persona está representada por la tensión fundamental, que es laque se mide primero en la mañana cuando la persona está aún descansada yantes de tomar alimentos.

Español18¿Qué es la hipertensión y cómo controlarla?
La hipertensión es la tensión arterial anormalmente alta, si no se corrige, puedecausar muchos problemas de salud, incluyendo ataques al corazón y apoplejía.
La hipertensión puede controlarse cambiando el estilo de vida, evitando elestrés y con medicamentos bajo la supervisión de un médico.
Para evitar la hipertensión o para mantenerla bajo control:
No fume.
Haga ejercicios con regularidad.
Reduzca la ingestión de sal y grasa.
Sométase a exámenes físicos con
Mantenga un peso apropiado.
regularidad.

¿Por qué debe medirse la tensión arterial en la casa?
La medición de la tensión arterial en una clínica o un consultorio médico puedecausar temor y, por lo tanto producir una lectura elevada, de 25 a 30 mmHg
mayor que la medida en la casa. La medición en casa reduce los efectos deinfluencias externas sobre las lecturas de tensión arterial, complementa laslecturas del médico y proporciona un historial más completo y exacto de latensión arterial.

Clasificación de tensión arterial según la OMS
OMS ha establecido las normas
para la evaluación de la tensión
arterial, tal como se muestra en el
gráfico.

Variaciones de tensión arterial
La tensión arterial de una persona
varía considerablemente en basediaria y estacional.
La misma
puede variar en 30 a 50 mmHg
dependiendo de diversas condiciones durante el día. En personas hipertensas,
estas variaciones son aun más notables.
La tensión arterial normalmente se eleva cuando se está trabajando o jugando
y baja a sus niveles más bajos al dormir. Por lo tanto, no se preocupe
demasiado de los resultados de una sola medición.
Mídase la tensión arterial a la misma hora
todos los días usando el procedimiento
descrito en este manual y tenga
conocimiento de su tensión normal.
Lecturas múltiples proporcionarán unhistorial de tensión arterial más completo.
Asegúrese de anotar la fecha yhora al registrar su tensión arterial.
Consulte con su médico para interpretar
los datos de su tensión arterial.

Español

19

Localización y corrección de fallos
Problema
No aparece nada
en la pantalla,
aun cuando seconecta laalimentación.

Causa probable
Las pilas se han agotado.
Los terminales de las pilas no
están en posición correcta.

El voltaje de las pilas está bajo.
Parpadea la marca
El brazalete no se (marca LOW BATTERY). Si laspilas están agotadas
infla.
completamente, no aparece
esta marca.
El brazalete no está ajustado
debidamente.

Acción recomendada
Reemplace todas las pilas
por otras nuevas.
Coloque las pilas con sus
terminales negativo y positivo
coincidiendo con losindicados en la banda inflable.
Reemplace todas las pilas
por otra nuevas.

Ajuste el brazaletecorrectamente.
Asegúrese de mantenerse
Movió su brazo o cuerpo
muy quieto y en silencio
durante la medición.
durante la medición.
El aparato no
Siéntese cómodamente ymide la presión
manténgase quieto. Coloquearterial. Las
el brazo sobre una mesa con
Elbrazaletenoestá
lecturas sonla palma hacia arriba y eldemasiado altas o posicionado correctamente.
brazalete al mismo nivel quedemasiado bajas.
el corazón.
Si el latido de su corazón es
muy débil o irregular, elaparato tendrá dificultad en
determinar su tensión arterial.
El valor se diferencia al medido Consulte el capítulo 3, ¿Poren una clínica o un consultorio qué debe medirse la tensión
arterial en la casa?
médico.
Otros
Retire las pilas. Colóquelas
nuevamente de forma
correcta e intente la
medición nuevamente.
Nota: Si las acciones descritas anteriormente no solucionan el problema, póngase
en contacto con su distribuidor. No abra el aparato ni intente repararlo, ya
que cualquier intento de reparación invalidará la garantía.

Español

20

Mantenimiento
No abra el aparato, ya que éste contiene componentes eléctricos delicados y
una unidad de aire compleja que podrían dañarse. Si no puede solucionar el
problema aplicando las instrucciones de localización y corrección de averías,
solicite el servicio de reparación a su proveedor o al grupo de servicio de A&D.
El grupo de servicio de A&D proveerá información técnica, repuestos yunidades a los proveedores autorizados.
Este aparato está diseñado y fabricado para una vida de servicio prolongada.
No obstante se recomienda someter el monitor a la inspección cada 2 años afin de asegurar el funcionamiento y la precisión adecuadas. Sírvase ponerse
en contacto con su distribuidor autorizado o A&D para su mantenimiento.

Datos técnicos
Tipo

UA-767S

Método de medición

Medición oscilométrica

Rango de medición

Tensión: 0 a 299 mmHg
Tensión sistólica: 60 a 279 mmHg
Tensión diastólica: 40 a 200 mmHg
Pulso:
40 a180 latidos/minuto

Precisión de medición

Tensión:Pulso:

Alimentación

4 pilas de 1,5V (R6P, LR6 o AA) o
un adaptador de CA (TB-233C) (No incluido))
Aprox. 700 veces LR6 (pilas alcalinas)
Aprox. 200 veces R6P (pilas de manganeso)
Con un valor de presión de 180 mmHg,
temperatura ambiente 23 ºC.

Número de mediciones±3 mmHg±5%

Clasificación

Equipo ME alimentado internamente (suministrado
por pilas) /Class II (suministrado por adaptador)
Modo de operación continuo

Prueba clínica

De acuerdo con ISO81060-2: 2013
En el estudio de validación clínica, se determinó lapresión arterial diastólica en 85 sujetos mediante
K5.

EMC

IEC 60601-1-2: 2014

Almacenamiento de datos

Las últimas 60 medidas

Condiciones ambientales
de funcionamiento
Condiciones de transporte

Temperatura de +10 a +40°C / RH de 15 a 85% /de 800 a 1060 hPa
Temperatura de -20 a +60°C / RH de 10 a 95 % /Español21

almacenamiento

Presión de 700 a 1060 hPa

Dimensiones

Aprox. 140 [An] x 60 [Al] x 105 [Pr] mm

Peso

Aprox. 245 g, excluyendozlas pilas

Protección de admisión

Aparato:Pieza aplicada

Brazalete Tip BF

Vida útil

Aparato: 5 años (cuando se usa seis veces al día)
Brazalete: 2 años (cuando se usa seis veces al día)
Adaptador de CA: 5 años (cuando se usa seis veces al día)

IP20

Accesorio: Adaptador de CA, El adaptador se conecta a la fuente de alimentación
modelo TB-233C
casera. Sírvase tener contacto con la agencia A&D
local para comprarlo.
Es necesario inspeccionar o sustituir el adaptador CA
periódicamente.
Símbolos impresos en el adaptador de CA
SímbolosFunción / Significado
Solo para uso en interiores
Aparato Clase II
Fusible térmico
FusibleEtiqueta de la directiva CE
Etiqueta de homologación EAC
Polaridad del enchufe del adaptador de CA
Accesorios vendidos por separado
BrazaleteNúmero de catálogo
CUF-F-LA
CUF-I
CUF-F-A

Adaptador de CA

Nota:

Tamaño del brazalete Tamaño del brazo
Brazalete grande de
31 cm a 45 cm
adultoBrazalete de amplio
22 cm a 42 cm
rango
Brazalete de adulto
22 cm a 32 cm
Tamaño del brazo: circunferencia en el bíceps.
Número de catálogo
TB-233C

Clavija
Tipo CLas especificaciones están sujetas a cambio sin previo aviso.
La clasificación IP hace referencia a los grados de protección otorgados por
los materiales contenedores según la norma CEI 60529. Este dispositivo está
protegido contra objetos extraños sólidos de 12 mm de diámetro y más, por
ejemplo, un dedo. Este dispositivo no está protegido contra el agua.

Español22s

IndiceIntroduzione .......................................................................... 2
Norme Precauzionali .............................................................. 2
Precauzioni ............................................................................ 2
Identificazione delle parti ...................................................... 5
Simboli .................................................................................. 6
Modalità operative ................................................................. 8
Preparazione del misuratore .................................................. 9
Installazione / Sostituzione delle batterie .......................... 9
Collegamento del bracciale allo strumento......................... 10
Collegamento dell’alimentatore a rete elettrica ............... 10
Come scegliere il bracciale corretto................................. 11
Applicazione del bracciale ............................................... 11
Consigli per l’esecuzione di misurazioni corrette ............ 12
Durante la misura ........................................................... 13
Dopo la misura ............................................................... 13
Misurazione della pressione ................................................. 14
Misurazione standard ..................................................... 14
Misurazione con impostazione della pressione di gonfiaggio desiderata.... 15
Note per una misurazione corretta .................................. 15
Richiamo delle misure memorizzate .................................... 16
Cosa è l’indicatore IHB/AFib? ............................................... 17
Cosa è l’AFib? ...................................................................... 17％IHB/AFib ........................................................................... 17
Indicatore dinamico della pressione ..................................... 18
Indicatore di classificazione dell’O.M.S................................. 18
Parlando di Pressione Arteriosa ............................................ 18
Cosa si intende per Pressione Arteriosa? ......................... 18
Cosa si intende per Ipertensione Arteriosa? ..................... 19
E come si controlla? ........................................................ 19
Perché è importante l’automisurazione al proprio domicilio? .... 19
Classificazione della pressione secondo le linee guida “O.M.S.” 19
Localizzazione guasti .......................................................... 20
Manutenzione ...................................................................... 21
Dati tecnici .......................................................................... 21

Italiano1Introduzione
Gentile Cliente, la ringraziamo per aver acquistato il Misuratore Elettronicodella Pressione Arteriosa A&D. Concepito per una facilità di uso ed alta
precisione, il nostro prodotto la aiuterà a controllare la pressione arteriosa
quotidianamente.
Raccomandiamo di leggere con cura questo manuale prima di utilizzare il
misuratore per la prima volta.

Norme Precauzionali
Il dispositivo è conforme alla Direttiva Europea CEE 93/42 per i Prodotti adutilizzo medico. Quanto sopra è evidenziato dal marchio di conformità.
(0123: Numero di riferimento per l'ente che riceve la notifica)
Il presente dispositivo deve essere utilizzato da parte di persone adulte, non dineonati o bambini.
Ambiente di utilizzo. Il dispositivo è idoneo per l’utilizzo da parte dell’utente inambienti di assistenza sanitaria domestici.
Il dispositivo è progettato per misurare la pressione sanguigna e la frequenza
cardiaca di soggetti umani a fini diagnostici.

Precauzioni
Il presente dispositivo è stato realizzato utilizzando componenti di precisione.
Si raccomanda di evitare condizioni estreme di temperatura, umidità, luce
solare diretta, urti o polvere.
Pulire il dispositivo e il bracciale utilizzando un panno morbido e asciutto oinumidito con acqua e detergente neutro. Evitare l’utilizzo di alcool, benzene,
solventi o altre sostanze chimiche aggressive per pulire il dispositivo o il
bracciale.
Evitare di piegare eccessivamente il bracciale o di riporre il tubo avvolto
strettamente per lunghi periodi, poiché quanto sopra potrebbe ridurre la
durata dei componenti.
Fare attenzione a evitare lo strangolamento accidentale di bambini o infanti
con il tubo e il cavo.
Durante la misurazione non attorcigliare il tubo prolunga, in quanto potrebbero
verificarsi delle lesioni dovute alla pressione continua nel bracciale.
Il dispositivo e il bracciale non sono impermeabili. Evitare il contatto con
pioggia, sudore e acqua.
Nel caso in cui il dispositivo sia utilizzato in prossimità di apparecchi televisivi, fornia microonde, telefoni cellulari, macchinari a raggi X o di altri dispositivi con forti
campi elettrici, le misurazioni potrebbero non risultare precise.
I dispositivi per la comunicazione wireless, come i dispositivi delle reti
domestiche, i telefoni cellulari, i telefoni cordless e le loro stazioni base, iwalkie-talkie possono interferire con il funzionamento di questo Monitor della
Pressione Sanguigna. Di conseguenza, è consigliabile mantenere una distanza
minima di 30 cm da tali dispositivi.
Italiano 2

Al riutilizzo del dispositivo, verificare che sia pulito.
Le apparecchiature, le parti e le batterie usate non devono essere trattatequali normali rifiuti domestici. Devono quindi essere smaltite in conformità alle
normative in vigore.
Se si utilizza l'alimentatore, assicurarsi che sia possibile scollegarlo dallapresa elettrica in caso di necessità.
Non modificare il dispositivo. Questa operazione può causare incidenti odanni al dispositivo stesso.
Per misurare la pressione sanguigna è necessario avvolgere strettamente il
bracciale intorno al braccio in modo da interrompere temporaneamente ilflusso sanguigno nell'arteria. Ciò può causare dolore, intorpidimento olasciare temporaneamente un segno rosso sul braccio, in particolare se la
misurazione viene ripetuta più volte di seguito. Il dolore, l'intorpidimento o isegni rossi che possono eventualmente verificarsi scompaiono dopo breve
tempo.
Misurazioni della pressione sanguigna troppo frequenti possono provocare
danni al paziente a causa delle interferenze con il flusso sanguigno. In caso di
uso ripetuto del dispositivo, assicurarsi che il funzionamento del misuratore
non comporti una prolungata compromissione della circolazione sanguigna.
Se si è stati sottoposti a mastectomia, rivolgersi a un medico prima diutilizzare il dispositivo.
Evitare che i bambini usino il dispositivo senza sorveglianza e tenere il dispositivo
fuori dalla loro portata. In caso contrario, si potrebbero causare incidente o danni.
Presenta piccole parti che possono causare soffocamento se ingerite
inavvertitamente dai bambini.
Scollegare l’adattatore C.a. quando non lo si usa durante la misurazione.
L’utilizzo di accessori non indicati in questo manuale potrebbe compromettere la
sicurezza.
In caso di cortocircuito della batteria, la suddette potrebbe riscaldarsi e causare
potenzialmente delle ustioni.
Consentire al dispositivo di adattarsi all’ambiente circostante, prima dell’uso (circa
un’ora).
Non sono stati condotti test clinici sui neonati e le donne in gravidanza. Non usare
su neonati o donne in gravidanza.
Non toccare contemporaneamente le batterie, la presa jack C.c. e il paziente. Si
potrebbero causare scosse elettriche.
Non gonfiare senza avvolgere il bracciale attorno al braccio.

Controindicazioni
Di seguito sono indicate le precauzioni da osservare per il corretto utilizzo deldispositivo.
Non applicare il bracciale a un paziente già collegato a un altro dispositivo
elettromedicale. Il dispositivo potrebbe non funzionare correttamente.
Le persone affette da grave deficit circolatorio al braccio devono consultare
un medico prima di utilizzare il dispositivo, al fine di evitare problemi medici.
Non eseguire auto-diagnosi dei risultati delle misurazioni, né iniziare
trattamenti in autonomia. Per la valutazione dei risultati ed eventuali
trattamenti, rivolgersi sempre al medico.
Italiano 3

Non applicare il bracciale se il braccio presenta ferite non cicatrizzate.
Non applicare il bracciale in caso di terapia con flebo o trasfusioni di sangue
eseguite al braccio del paziente. Pericolo di lesioni o incidenti.
Non utilizzare il dispositivo in ambienti in cui siano presenti gas infiammabili,
quali i gas anestetici. Pericolo di esplosione.
Non utilizzare il dispositivo in ambienti con concentrazioni molto elevate diossigeno, quali camere iperbariche o tende a ossigeno. Pericolo di incendio oesplosione.

Italiano 4

Identificazione delle parti
Display a cristalli liquidi

Spinadell’alimentat

Spina innesto aria

Jack a CC
Pulsante START(AVVIO)

Presa d’aria

Indicatore di scelta delbracciale adeguato

Tubo prolunga
Simbolo di indicazione
posizione arteria brachiale

Indicatore di posizione
Bracciale
Vano batterie
Coperchio vano batterie

Batterie da 1,5 V(R6P, LR6 o AA)

Particolari del Display

Valori medi

MEMORIA
Simbolo IHB/AFib
Pressione
sistolica%IHB/AFib

Simbolo di errore diposizionamento del bracciale

Pressione
diastolica

Simbolo di errore di movimento
Simbolo del cuore(Indicazione misura in corso)

Frequenza
cardiaca

Indicatore di classificazione OMS e
indicatore di pressione
Indicatore batterie

Italiano 5

Simboli
Simboli impressi sulla confezione del dispositivo
Simboli
SYS
DIA
PUL

EC

REP

Funzione/Significato
Posizionare l’apparecchio in modalità standby, quindi collegarlo
Pressione arteriosa sistolica in mmHg
Pressione arteriosa diastolica in mmHg
Battiti cardiaci al minuto
Guida all'installazione delle batterie
Corrente continua
Tipo BF: dispositivo, bracciale e tubi sono stati progettati al fine digarantire una protezione particolare contro eventuali scosse
elettriche.
Etichetta Direttiva CE sui dispositivi medici
Rappresentante UE
Produttore
Data di produzione

IP

SN

Simbolo internazionale di protezione
Etichetta Direttiva RAEE (direttiva Europea di smaltimento dei
dispositivi elettrici ed elettronici)
Numero di matricola
Consultare il manuale/l'opuscolo di istruzioni
Polarità del jack a CC
Tenere asciutto

Simboli visualizzati sul display
SimboliFunzione/Significato
Azione raccomandata
Indicazione visualizzata durante la fase La misurazione è in corso.
di misurazione. Lampeggia durante il
Rimanere quanto più
rilevamento della frequenza cardiaca.
possibile fermi e rilassati.
Simbolo IHB/AFib
Indicazione visualizzata se viene
rilevato un battito cardiaco irregolare.
Potrebbe illuminarsi se viene rilevata
una leggerissima vibrazione come unbrivido o un tremore.
La lettura potrebbe
presentare un valore
incorretto.
Simbolo visualizzato se viene rilevato
un movimento corporeo o del braccio. Ripetere la misurazione.
Rimanere quanto possibile
fermi e rilassati.
Italiano 6

Simboli
SimboliFunzione/Significato

Simbolo visualizzato in fase dimisurazione se il bracciale è allentato.

Azione raccomandata
La lettura potrebbe
presentare un valore
incorretto.
Applicare e stringere
adeguatamente il bracciale
e ripetere la misurazione.

Frequenza rilevata di IHB/AFib in memoria
Numero di IHB/AFibrilevati in memoria
x 100 [%]
%IHB/AFib =Numero totale
Misurazioni precedenti
MEMORIZZATE
Valori medi
BATTERIE CARICHE

Indicatore di stato di carica dellebatterie durante la misurazione
BATTERIE SCARICHE

Indicatore di batterie in esaurimento
Valori della pressione non corretti acausa di movimenti durante la
misurazione
Differenza tra pressione sistolica ediastolica inferiore a 10 mmHg
Pressione di gonfiaggio non sufficiente
Il bracciale non è applicato in modo
corretto
ERRORE VISUAL. POLSO

Qualora l’indicatore
lampeggi, sostituire tutte lebatterie.
Ripetere la misurazione.
Rimanere quanto possibile
fermi e rilassati.

Applicare e stringere
adeguatamente il
bracciale e ripetere la
misurazione.

Il battito cardiaco non è stato rilevato inmodo corretto

Errore interno del misuratore dipressione arteriosa

Italiano 7

Rimuovere le batterie epremere il pulsante
START (AVVIO), quindi
reinstallare le batterie. Se
l’errore persiste, contattare
il rivenditore.

Modalità operative
1. Misurazione normale con memorizzazione dei dati
Premere il pulsante START . La pressione arteriosa viene misurata e i dati sono
memorizzati nel dispositivo. Sono memorizzati automaticamente gli ultimi 60
valori di pressione.

2. Richiamo dei dati
Quando non è visualizzato niente, tenere
premuto il pulsante _START_.

In standby,
tenere premuto il pulsante
START

Rilasciare il pulsante quando si visualizza la
media dei dati.

Vengono visualizzati automaticamente il
numero dei dati e il numero dei dati
memorizzati a partire dall'ultima
misurazione.

Per i dettagli su come richiamare i dati,
consultare “Richiamo delle misurememorizzate”.

3. Per eliminare tutti i dati memorizzati
Durante lo spegnimento del dispositivo, premere e tenere
premuto il pulsante _START_ (AVVIO) sino a visualizzare la
scritta "".
" per cancellare i dati.
Selezionare "lampeggia.
I dati sono cancellati quando il simbolo
Il dispositivo si spegne automaticamente.

START

4. Misurazione con impostazione della pressione di gonfiaggio
desiderata
Vedere descrizione completa a pagina 15.
Italiano 8

Preparazione del misuratore
Installazione / Sostituzione delle batterie
1. Rimuovere il coperchio del vano batterie.
2. Rimuovere le batterie esaurite ed inserire
le nuove batterie, rispettando le polarità(+ e -) come indicato nella figura a lato.
Utilizzare esclusivamente batterie R6P,
LR6 o AA.

Procedura 1
Procedura 3

Procedura 1

Procedura 2

3. Chiudere il coperchio del vano batterie.

ATTENZIONE
Inserire le batterie come indicato nel vano porta
batterie. In caso contrario, l'apparecchio non
funzionerà.
Quando il simbolo(BATTERIE SCARICHE) lampeggia sul display, sostituire
tutte le batterie. Non mischiare batterie nuove con batterie usate, per evitare diridurre la durata delle batterie o causare malfunzionamenti del dispositivo.
Spento il dispositivo, attendere almeno due secondi per sostituire le batterie.
(BATTERIE SCARICHE) viene visualizzato anche dopo averSe il simbolo
sostituito le batterie, eseguire una misurazione della pressione arteriosa. Il
dispositivo è in seguito in grado di riconoscere le batterie nuove.
Il simbolo(BATTERIE SCARICHE) non viene visualizzato in caso di batteriecompletamente esaurite.
La durata delle batterie varia in base alla temperatura ambiente e potrebbe
risultare ridotta in caso di esposizione a basse temperature. In genere, quattro
batterie LR6 nuove durano circa un anno, quando vengono utilizzate due volte ogni
giorno per la misurazione.
Utilizzare solamente le batterie del tipo specificato. Le batterie fornite indotazione al dispositivo sono a solo scopo di prova delle prestazioni del monitor ela loro durata potrebbe essere breve.
Togliere le batterie nel caso in cui il dispositivo non sia utilizzato per lungo tempo.
Le batterie potrebbero infatti essere soggette a perdite, causando quindi dei
malfunzionamenti.

Italiano 9

Preparazione del misuratore
Collegamento del bracciale allo
strumento

Presa d’aria

Inserire fermamente la spina innesto
aria nella presa d’aria.

Spina innesto aria

Collegamento dell’alimentatore a rete elettrica
Inserire la spina dell’alimentatore nel jacka CC.
Inserire successivamente l’alimentatore
nella presa elettrica.

Jack a CC

Utilizzare l’alimentatore a rete elettrica
specificato.( Consultare pagina 22.)
Per scollegare l’alimentatore dalla presa
elettrica, afferrare e rimuovere il corpo
dell’alimentatore dalla presa.
Per scollegare la spina dell’alimentatore dal
Misuratore Elettronico Della Pressione
Arteriosa, afferrare ed estrarre la spinadell’alimentatore dal misuratore.

Italiano

10

Spina dell’alimentatore

Preparazione del misuratore
Come scegliere il bracciale corretto
L’uso della dimensione corretta di un bracciale è importante per una lettura
precisa. Se il bracciale non presenta la dimensione corretta, la lettura potrebbe
produrre un valore incorretto della pressione sanguigna.
La dimensione del braccio è stampata su ogni bracciale.
Il simbolo
e l'indicatore di scelta del bracciale adeguato determinano se è
stato utilizzato il bracciale corretto. (Consultare "Simboli stampati sul bracciale "alla pagina seguente.)
Se il simbolo
è posizionato fuori dal range, si raccomanda l’acquisto di un
bracciale appropriato (consultare la tabella sottostante).
Il bracciale è un prodotto di normale consumo. In caso di usura, acquistarne uno nuovo.
Dimensione del braccio
31 cm a 45 cm
22 cm a 42 cm
22 cm a 32 cm

Dimensione raccomandata delbracciale
Bracciale grande per adulti
Bracciale ampio
Bracciale per adulti

Numero di catalogo
CUF-F-LA
CUF-I
CUF-F-A

Misura del braccio: corrisponde alla circonferenza del bicipite.
Nota:
il modello UA-767S non è indicato per l'uso con il bracciale
piccolo per adulti.

Applicazione del bracciale
1. Avvolgere il bracciale intorno al
braccio superiore, a circa 1 - 2 cm
sopra la piega del gomito, comeindicato nella figura a lato. Applicare il
bracciale direttamente a contatto conla pelle, in quanto i vestiti possono
causare errori nella misura.

Arrotolare la manica evitando
di stringere troppe il braccio

Bracciale
1-2 cmTubo prolunga
Simbolo di indicazione
posizione arteria brachiale

2. La compressione al braccio causata
da un'avvolgimento troppo strettodella manica della camicia, puo`
causare una misura errata.
3. Verificare che il simbolo
all'interno del range di selezione.

Avvolgere il bracciale
saldamenteèSimbolo

Italiano

11

e gamma adatta adeguataPreparazione del misuratore
Simboli stampati sul bracciale
Simboli●▲
REF

Funzione/Significato

Azione raccomandata

Simbolo di indicazione posizione arteria
brachiale

Posizionare il simbolo ●sull’arteria del braccio o inallineamento con il dito anulare
nella parte interna del braccio.

IndiceNumero di catalogo
Gamma adatta a bracciale per adulti.
È stampato sul bracciale per adulti.
Gamma superiore stampata sul bracciale Utilizzare il bracciale grande
per adulti anziché il bracciale
per adulti e sul bracciale ampio.
per adulti o il bracciale ampio.
Gamma adatta a bracciale ampio.
È stampato sul bracciale ampio.
Gamma adatta a bracciale grande per
adulti.
È stampato sul bracciale grande per adulti.
Gamma inferiore stampata sul bracciale per
adulti e sul bracciale ampio.
il bracciale per adulti
Gamma inferiore stampata sul bracciale Utilizzare
anzichéilbracciale grande per
grande per adulti.
adulti.
Numero di lotto
Indicatore di scelta del bracciale adeguato

Bracciale grande per adulti

AL

Bracciale ampio

S

W

L

Bracciale per adulti

S

A

L

Consigli per l’esecuzione di misurazioni correttePer garantire la migliore precisione delle rilevazioni suggeriamo di:
Sedersi comodamente su una sedia. Appoggiare il braccio sul tavolo. Non
incrociare le gambe. Tenere i piedi appoggiati al suolo e raddrizzare il dorso
Prima di procedere alla misurazione rilassarsi per 5 -10 minuti.
Avvolgere il bracciale allo stesso livello del cuore.
Durante la misurazione, mantenersi fermi e tranquilli.
Non effettuare la misurazione subito dopo aver eseguito esercizi fisico o aver
fatto il bagno. In tali casi, riposare per venti-trenta minuti prima di procedere alla
misurazione.
Si consiglia di misurare la pressione arteriosa tutti i giorni alla stessa ora.
Italiano12Preparazione del misuratore
Durante la misura
Durante la misurazione, è normale sentire il bracciale a pressione stringere
fortemente (Non preoccuparsi).

Dopo la misura
Dopo la misurazione, premere il pulsante di START per spegnere il dispositivo.
Rimuovere il bracciale e registrare i vostri dati.
Nota: L'apparecchio
prevedeuna
funzioned'interruzione
automaticadell'alimenta-zione, che disattiva automaticamente l'alimentazione, un minuto
dopo la misurazione.
E’ consigliabile lasciar passare almeno 3 minuti prima di effettuare una nuova
misura sulla stessa persona.

Italiano13

Misurazione della pressione
Prima di eseguire la misurazione, consultare “Note per una misurazione
corretta“ alla pagina seguente.

Misurazione standard
1.

Applicare il bracciale preferibilmente sul braccio sinistro
allo stesso livello del cuore. Durante la misurazione
rimanere quanto più possibile fermi e rilassati.

2.

Premere il pulsante START .
Per un attimo verranno visualizzati tutti i segmenti
sul display e in un secondo tempo, come mostrato
nella figura a lato, verrà visualizzato lo 0 (zero)lampeggiante per alcuni istanti.
Successivamente il bracciale inizierà a gonfiarsi(è normale sentire stringere decisamente) e unindicatore dinamico a barre sarà visualizzato
nell'area a sinistra del display, mostrandol’incremento della pressione di gonfiaggio.
Nota: Nel caso in cui, in qualsiasi momento, si
desideri sospendere la misurazione,
premere nuovamente il pulsante START .

3. Una volta completata la fase di pressurizzazione
lo sgonfiamento sarà automatico ed il simbolo(simbolo del cuore) inizierà a lampeggiare,
indicando che la misurazione è in corso. Una
volta rilevate le pulsazioni, il simbolo del cuore
lampeggerà in sincronia con il battito cardiaco.
Nota: Nel caso in cui non si ottenga una pressione
di gonfiaggio adeguata, il misuratore inizieràautomaticamente a rigonfiare il bracciale.
Per evitare il rigonfiaggio, consultare“Misurazione con impostazione dellapressione di gonfiaggio desiderata” alla
pagina successiva.
4.

5.

Una volta completata la misurazione, saranno
visualizzati i valori relativi alla pressione
sistolica, diastolica e alla frequenza cardiaca,
insieme a un segmento lampeggiante incorrispondenza della scala dei valoridell’O.M.S. Il bracciale espellerà l’aria residua
e si sgonfierà completamente.
Premere il pulsante START per spegnere il
dispositivo.

Nota:

E’ consigliabile lasciar passare almeno 3 minuti prima
di effettuare una nuova misura sulla stessa persona.

Italiano

14

Al livello delcuore
PremereSTART

Tutte le parti deldisplay vengono
visualizzate

Visualizzazione
zero
Inizio della fase digonfiamento

Pressurizzazione

Misurazione
in corso

Pressione sistolica
Pressione diastolica
Classificazione OMS
Frequenza cardiaca
Espulsione automatica
dell’aria residua nel braccialeMisurazione della pressione
Misurazione con impostazione della pressione di gonfiaggio desiderata
Il modello UA-767S è stato progettato per rilevare in modo completamente
automatico la pressione e la frequenza cardiaca.
Applicare questo metodo se il gonfiaggio si ripete, o se i risultati non vengono
visualizzati malgrado la riduzione della pressione a 20 mmHg o meno.
1.

Avvolgere il bracciale sul braccio(preferibilmente il sinistro) all'altezzadel cuore.

A livello del cuore
Premere il pulsante _START
START_

2.

Premere il pulsante _START_ (AVVIO).

3.

Mentre lo zero lampeggia, premere
e tenere premuto il pulsante_START_ (AVVIO) sino alla
visualizzazione di un valore
superiore di 30 - 40 mmHg rispetto
Premere e tenere premuto il START
alla pressione sistolica prevista.

Alla visualizzazionedella pressione digonfiaggio richiesta,
rilasciare il pulsantepulsante START

4.

Dopo aver raggiunto il valore
desiderato, rilasciare il pulsante_START_, la misurazione inizierà.
Proseguire quindi con la
misurazione della pressione
sanguigna come descritto alla
pagina precedente.
Per quanto riguarda la procedura di
rilevazione della pressione, fareriferimento alla pagina precedente

Note per una misurazione corretta
Sedersi in una posizione comoda. Appoggiare il braccio su un tavolo con il palmodella mano rivolto verso l’alto e con il bracciale alla stessa altezza del cuore.
Rilassarsi per circa 5 - 10 minuti prima di effettuare una misurazione. Se si èagitati o emozionati, la misurazione rifletterà questa situazione, con un valoredecisamente più alto (o più basso) della pressione arteriosa normale. Anche lafrequenza cardiaca sarà solitamente più alta del normale.
La pressione arteriosa di un soggetto varia costantemente, secondo l’attività
fisica ed emozionale, l’assunzione di cibi e la luce. Anche l’assunzione dibevande alcoliche può avere un effetto sull’andamento dei valori di pressione
arteriosa.
Questo apparecchio si basa sul principio di misura oscillometrico, quindi ingrado di valutare e analizzare i parametri delle pulsazioni cardiache. Se i battiti
sono molto deboli o irregolari, l’apparecchio potrebbe avere difficoltà adeterminare la pressione arteriosa.
Italiano15

Se l'apparecchio rileva una circostanza anormale, arresterà la misurazione e
visualizzerà un simbolo di errore. (Vedere pagina 7 per la descrizione dei simboli).
Questo dispositivo deve essere gestito da persone adulte. Consultare il proprio
medico prima di utilizzare lo strumento su un bambino. In ogni caso, l’utilizzo su
bambini deve avvenire sempre sotto la stretta sorveglianza di persone adulte.
La prestazioni del monitor automatico per la pressione arteriosa potrebbero
essere influenzata dalla temperatura o umidità eccessiva o l'altitudine.

Richiamo delle misure memorizzate
Nota: Il dispositivo memorizza automaticamente gli ultimi 60 valori di pressione.
1.

Quando non è visualizzato alcun
elemento, premere e tenere
premuto il pulsante _START_(AVVIO) per richiamare i dati
memorizzati.

2.

Rilasciare il pulsante quando si
visualizza la media dei dati.

3.

Vengono visualizzati
automaticamente il numero dei dati
e il numero dei dati memorizzati apartire dall'ultima misurazione.

4.

Il display si spegne
automaticamente dopo lavisualizzazione di tutti i dati.

Premere e tenere
premuto il pulsante STARTSTARTMedia ditutti i dati

Pressione sistolica
media
Pressione diastolica
media
Frequenza cardiaca
media

Sistolica
Diastolica
Frequenza
cardiaca

Nota: se si preme il pulsante _START_(AVVIO) durante il richiamo dei
dati, il dispositivo si spegne.

Dati più recenti

Sistolica
Diastolica
Frequenza
cardiaca
Ultimi dati (Dati più vecchi)
Italiano

16

Cosa è l’indicatore IHB/AFib?
Quando il monitor rileva un ritmo irregolare durante le misurazioni, l’indicatore
IHB/AFib sarà visualizzato sul display unitamente ai valori di misurazione.
Nota: Raccomandiamo di contattare il proprio medico se viene indicato
frequentemente questo simbolo IHB/AFib.

Cosa è l’AFib?
Il cuore si contrae a causa di segnali elettrici che hanno luogo nel cuore e invia
sangue in tutto il corpo. La fibrillazione atriale (AFib) si verifica quando ilsegnale elettrico nell’atrio diventa confuso e causa disturbi nell’intervallo dellepulsazioni. L’AFib può causare ristagni del sangue nel cuore, i quali possono
causare facilmente coaguli di sangue, una delle cause di ictus e infarto.

％IHB/AFib%IHB/AFib è visualizzato come frequenza delI’IHB/AFib rilevato.
L’IHB/AFib può rilevare non soltanto i rumori come il movimento fisico, maanche un battito cardiaco irregolare. Si consiglia pertanto di consultare il
proprio medico se il livello di %IHB/AFib è elevato.
Numero di IHB/AFibrilevati in memoria
x 100 [%]
%IHB/AFib =Numero totale
Visualizzazione di %IHB/AFib: %IHB/AFib viene visualizzato quando si
visualizzano i valori medi. (Consultare “2.
Richiamo dei dati” in “Modalità operative”)%IHB/AFib non è visualizzato quando il numero
della memoria è inferiore o pari a sei.
Visualizzazione valori medi

Livello 0%IHB/AFib=0

Livello 1%IHB/AFib=1 - 9

Livello 2
Livello 3%IHB/AFib=10 - 24 %IHB/AFib=25 - 100

Non visualizzato

Italiano17

Indicatore dinamico della pressione
Lo strumento è provvisto di un indicatore dinamico a barre che controlla il progressodella pressione durante la fase di misurazione. L’indicatore è visibile nella partesinistra del display e, si incrementa e diminuisce progressivamente durante il
gonfiaggio del bracciale e durante la misura della pressione.
Pressurizzazione
Depressurizzazione

Gonfiamento in corso

Misurazione in corso
Gonfiamento

Indicatore di classificazione dell’O.M.S.
L’Organizzazione Mondiale della Sanità (O.M.S.)
ha sancito delle linee guida per la corretta
valutazione e interpretazione dei valori di pressione
misurata.
Ha suddiviso in 6 stadi la classificazione dei valori,come evidenziato dalla tabella sottostante:

Indicatore di classificazione O.M.S.

Esempio:

Ipertensione moderata

Ipertensione severa (grado 3)
Ipertensione moderata (grado 2)
Ipertensione lieve (grado 1)
Pressione normale – alta
Pressione normale
Pressione ottimale

Ipertensione lieve: L’indicatore mostra un segmento sulla base dei
dati rilevati, corrispondente alla
Classifica zione O.M.S

Pressione normale – alta

Parlando di Pressione Arteriosa
Cosa si intende per Pressione Arteriosa?
La pressione arteriosa è la forza esercitata dal sangue contro le pareti delle arterie.
La pressione sistolica (massima) si verifica nel momento in cui il cuore si contrae espinge il sangue nelle arterie. La pressione diastolica (minima) si verifica quando il cuore
si dilata, ricevendo il sangue all’interno del ventricolo sinistro, e si prepara a una nuova
contrazione. La pressione arteriosa è misurata in millimetri di mercurio (mmHg).
Italiano18

Cosa si intende per Ipertensione Arteriosa?
Secondo le linee guida dell’Organizzazione Mondiale della Sanità, si può parlare diipertensione arteriosa quando si hanno valori di pressione sistolica e diastolica
persistentemente elevati oltre il limite di 140 / 90 mmHg. E’ importante però non fareiferimento esclusivamente ad una singola misura, ma è necessario considerare anche ilvalore di più rilevazioni in un periodo di tempo stabilito dal medico ed anche lefluttuazioni della pressione stessa. Recenti studi a lungo termine hanno dimostrato che ilimiti da considerare per la pressione domiciliare automisurata, sono infatti meno elevati
e si attestano su valori di 135 mmHg per la pressione sistolica e 85 mmHg per lapressione diastolica.

E come si controlla?
E’ possibile tenere sotto controllo l’ipertensione arteriosa variando il proprio stile di vita ealimentare, evitando il più possibile situazioni di stress, oppure con opportuni trattamenti
farmacologici secondo prescrizione medica. Per prevenire l'ipertensione arteriosa o pertenerla sotto controllo, le linee guida dell’ O.M.S. consigliano di:Non fumare
Eseguire esercizi a cadenza regolare
Ridurre l’uso di sale e moderare
Sottoporsi a check-up regolaril’assunzione di grassi animali
Mantenere il proprio peso a livelli ottimali

Perché è importante l’automisurazione al proprio domicilio?
La pressione che il medico misura mediante lo sfigmomanometro tradizionale è quasi
sempre caratterizzata da un rialzo generato dall’emozione che colpisce il paziente al
momento della visita medica, rialzo la cui entità e durata sono variabili da soggetto asoggetto. Questo fattore è comunemente conosciuto come ”ipertensione da camicebianco” o più opportunamente come “ipertensione clinica isolata” e si manifesta a causa
degli stati di ansia generati dalla presenza del medico o della struttura sanitaria.
Un vantaggio della misurazione domiciliare è che in genere è esente da questo fenomeno.
Un altro vantaggio dell’automisurazione è che permette di ottenere più valori nell’arco didiversi giorni o settimane, con una stima precisa delle variazioni della pressione.
La misurazione domiciliare inoltre può fornire utili informazioni nella valutazione deltrattamento antiipertensivo. In questo caso la pressione domiciliare può rivelarsi un
metodo molto efficace, perché il suo impiego coinvolge attivamente il paziente con effetti
potenzialmente positivi sull’aderenza alla terapia.

Classificazione della pressione secondo le linee guida “O.M.S.”
Gli standard per la valutazione della pressione
arteriosa, indipendentemente dall’età, sono stati
stabiliti dall’Organizzazione Mondiale della Sanità(O.M.S.), come mostrato nella tabella a fianco.
Normalmente, la pressione arteriosa aumenta
durante l’attività quotidiana e il lavoro e, tende adiminuire durante il sonno. Le linee guida inoltre
consigliano di effettuare quotidianamente lemisurazioni possibilmente sempre alla stessa ora.
L’esecuzione di molte rilevazioni permette di ottenere un quadro della propria pressione
arteriosa più completo. E’ sempre meglio annotare la data e l’ora della registrazionedella pressione arteriosa al fine di favorire una miglior diagnosi da parte del proprio
medico curante.
Italiano19Raccomandiamo di consultare sempre il proprio medico
al fine di interpretare correttamente i dati relativi allapressione.

Localizzazione guasti
Problema

Causa probabile

Misura correttiva
Sostituire le batterie usate con
Le batterie sono scariche.
Sul display, nonbatterie nuove.
viene visualizzato
Le batterie sono state inserite Posizionare le batterie in modo talenessun messaggio,
non correttamente.
che i poli positivo e negativo
anche dopo
I poli delle batterie sono
coincidano con quelli indicati nell’accensione.
posizionati in modo errato.
vano batterie.
Il voltaggio delle batterie è
troppo basso. il simbolo(BATTERIE SCARICHE)Il bracciale non si
Sostituire le batterie usate con
lampeggia. In caso di batteriegonfia.
batterie nuove.
completamente scariche, ilsimbolo non saràvisualizzato.
Il bracciale non è avvolto in
Avvolgere il bracciale in modo
modo adeguato.
adeguato.
Durante la fase di
Durante la fase di misurazione,
misurazione, avete mosso ilrimanere fermi e rilassati.
braccio o il corpo.
Il dispositivo nonSedere comodamente e rimanere
esegue lafermi e rilassati. Appoggiare il braccio
misurazione.
La posizione del bracciale
su un tavolo con il palmo della mano
I valori ottenuti sono non è corretta.
rivolto verso l’alto e con il bracciale alla
eccessivamente alti
stessa altezza del cuore.
o bassi.
Nel caso in cui il battito cardiaco
sia molto debole o irregolare, il
dispositivo potrebbe trovaredifficoltà nel determinare lapressione arteriosa.
Il valore al domicilio è diverso Far riferimento al capitolo “Perchéda quello misurato in clinica o è importante l’automisurazione alpresso lo studio medico.
proprio domicilio?”
Vari
Rimuovere le batterie. Posizionarlecorrettamente e eseguirenuovamente la misurazione.
Nota: Nel caso in cui le azioni sopra riportate non risolvano il problema, contattare il
rivenditore. Non tentare di aprire o riparare questo prodotto autonomamente, perchési rischia di invalidare la garanzia.

Italiano

20

Manutenzione
Non aprire il dispositivo che include componenti elettrici delicati ed un sofisticato
impianto ad aria che potrebbero danneggiarsi. Nel caso in cui non si riesce a risolvere il
problema anche dopo aver eseguito quanto riportato nelle istruzioni di cui al capitolo“Localizzazione guasti”, contattare il centro di assistenza più vicino, oppure rivolgersi al
rivenditore o al punto vendita dal quale ha acquistato il misuratore.
Il dispositivo è stato concepito e fabbricato per una lunga durata di servizio. Tuttavia si
raccomanda di controllare lo strumento ogni 2 anni, per garantire funzionamento e
precisione attendibili.

Dati tecnici
Tipo

UA-767S

Metodo di misurazione

Oscillometrico

Range di misurazione

Pressione: 0 - 299 mmHg
Pressione sistolica: 60-279 mmHg
Pressione diastolica: 40-200 mmHg
Pulsazioni: 40 - 180 battiti/minuto

Precisione

Pressione: ±3 mmHg
Pulsazioni: ±5%
4 batterie da 1,5V (R6P, LR6 o AA) o
Alimentatore a rete elettrica (TB-233C) (non incluso)
Circa 700
LR6 (batterie alcaline)
Circa 200
R6P (batterie al manganese)
A una pressione di 180 mmHg, temperatura ambiente di
23 °C.
Dispositivo medico elettronico ad alimentazione interna(alimentazione con batterie) /Class II (alimentazione con alimentatore a rete)
Modalità di funzionamento continuo

Alimentazione
Numero di misurazioni

Classificazione

Test clinici

In conformità ad ISO81060-2：2013
Nello studio di convalida clinica, K5 è stato utilizzato
su 85 soggetti per la determinazione della pressione
arteriosa diastolica.

EMD
Memoria dati
Condizioni d’esercizio

IEC 60601-1-2: 2014
Le ultime 60 misurazioni
da +10° a +40°C/ da 15% a 85% RH /da 800 a 1060 hPa
da -20° a + 60°C /da 10% a 95% RH /Condizioni di trasporto /conservazione
Dimensioni
Peso
Protezione contro
Parte applicata

da 700 a 1060 hPa

Circa 140 [L] x 60 [A] x 105 [P] mm
Circa 245 g, batterie escluse
Dispositivo: IP20
Bracciale Tipo BF
Italiano21

Vita utile

Dispositivo: 5 anni (se utilizzato sei volte al giorno)
Bracciale : 2 anni (se utilizzato sei volte al giorno)
Alimentatore: 5 anni (se utilizzato sei volte al giorno)

Alimentatore opzionale
L'alimentatore ha lo scopo di collegare il monitor della pressione sanguigna a una fonte dialimentazione in ambiente domiciliare. Rivolgersi al rivenditore A&D di zona per l'acquisto.
L'alimentatore deve essere ispezionato o sostituito periodicamente.

Alimentatore opzionale

TB-233C

L'alimentatore ha lo scopo di collegare il monitor dellapressione sanguigna a una fonte di alimentazione inambiente domiciliare.
Per l’acquisto, contattare il punto vendita autorizzato
A&D.
L'alimentatore a rete elettrica deve essere ispezionato
o sostituito periodicamente.
Simboli riportati sull’alimentatore
SimboliFunzione / Significato
Per l'uso solo in ambienti interni
Dispositivo di Classe II
Fusibile termico
Fusibile
Etichetta del dispositivo relativa alla
direttiva CE
Etichetta del dispositivo relativa alla
certificazione EAC
Polarità dello spinotto dell'alimentatore

Accessori venduti separatamente
Numero di catalogo Dimensione del bracciale Dimensione del braccio
Bracciale grande per adulti
CUF-F-LA
31 cm a 45 cm
CUF-I
Bracciale ampio
22 cm a 42 cm
Bracciale per adulti
CUF-F-A
22 cm a 32 cm
Determinazione circonferenza del braccio: si misura a metà tra spalla e gomito.

Bracciale

Alimentatore a rete elettrica

Nota:

Numero di catalogo
TB-233C

Spina
Tipo CLe specifiche sono soggette a variazione, senza alcun preavviso.
La classificazione IP indica i gradi di protezione degli involucri in conformità con lanorma CEI 60529. Questo dispositivo è protetto contro corpi solidi estranei di 12mm di diametro o più grandi, come per esempio un dito. Questo dispositivo non è
protetto contro i getti d’acqua.

Italiano22Índice
Estimados clientes ................................................................ 2
Observações preliminares ..................................................... 2
Precauções ........................................................................... 2
Identificação de componentes............................................... 5
Símbolos .............................................................................. 6
Modo de funcionamento ....................................................... 9
Utilização do dispositivo de monitorização ......................... 10
Instalação/substituição das pilhas ................................. 10
Ligação do tubo de ar .................................................... 11
Ligação do adaptador de CA .......................................... 11
Seleção do tamanho correto da braçadeira ..................... 12
Aplicação da braçadeira ................................................. 12
Como efetuar medições precisas .................................... 14
Medição ......................................................................... 14
Após a medição ............................................................. 14
Medições ............................................................................ 15
Medição normal ............................................................. 15
Medição com a tensão sistólica desejada ....................... 16
Notas para medições corretas ........................................ 16
Revisão dos dados da memória ........................................... 17
O que é o indicador de IHB/AFib? ........................................ 18
A que se refere AFib? .......................................................... 18％IHB/AFib .......................................................................... 18
Indicador da barra de tensão .............................................. 19
Indicador de classificação segundo a OMS .......................... 19
Sobre tensão arterial ........................................................... 19
O que é a tensão arterial?............................................... 19
O que é a hipertensão e como se controla? .................... 20
Porque deve medir a tensão arterial em casa? ................ 20
Classificação da tensão arterial segundo a OMS ............. 20
Variações da tensão arterial ........................................... 20
Resolução de problemas ..................................................... 21
Manutenção ........................................................................ 22
Dados técnicos ................................................................... 22

Português 1

Estimados clientes
Parabéns pela aquisição de um dispositivo de monitorização de tensão
arterial A&D de última geração. Desenvolvido para facilidade deutilização e precisão, este dispositivo irá facilitar a sua rotina diária na
medição da tensão arterial.
Recomendamos que leia atentamente o presente manual de
instruções antes de utilizar o dispositivo pela primeira vez.

Observações preliminares
Este dispositivo encontra-se em conformidade com a Diretiva Europeia
93/42 CEE para produtos médicos. Tal é evidenciado pela
marca
de conformidade. (0123: o número de referência ao organismo notificado
envolvido)
O dispositivo foi desenvolvido para utilização em adultos, não devendo
ser utilizado em recém-nascidos ou bebés.
Âmbito de utilização. O dispositivo destina-se à utilização autónoma porparte do utilizador para cuidados de saúde em ambiente doméstico.
Este dispositivo foi desenvolvido para medir a tensão arterial e afrequência de pulsação de pessoas para fins de diagnóstico.

Precauções
A construção deste dispositivo inclui componentes de precisão. Devem
evitar-se extremos de temperatura, humidade, luz solar direta, choque ou
poeira.
Limpe o dispositivo e a braçadeira com um pano seco e macio ou com
um pano humedecido com água e um detergente neutro. Nunca utilizeálcool, gasolina, diluente ou outros produtos químicos agressivos para
limpar o dispositivo ou a braçadeira.
Evite dobrar a braçadeira apertada ou guardar o tubo torcido por
períodos prolongados, uma vez que tal pode encurtar a vida útil dos
componentes.
Tenha cuidado para evitar o estrangulamento acidental de bebés ou
crianças com o tubo e o cabo.
Não torça o tubo de ar durante a medição. Caso contrário, podem
resultar lesões devido à pressão contínua da braçadeira.
O dispositivo e a braçadeira não são resistentes à água. Evite que achuva, a transpiração e a água molhem o dispositivo e a braçadeira.
As medições podem sofrer distorções se o dispositivo for utilizado
próximo de televisores, micro-ondas, telemóveis, raios-X ou outros
dispositivos com fortes campos elétricos.
Português 2

Os dispositivos de comunicação sem fios, tais como os dispositivos derede doméstica, telemóveis, telefones sem fios e as respetivas centrais ewalkie-talkies, podem afetar o dispositivo de monitorização de tensão
arterial. Como tal, é necessário manter uma distância mínima de 30 cm
entre esses dispositivos.
Quando reutilizar o dispositivo, confirme se está limpo.
O dispositivo, os respetivos componentes e as pilhas usados não são
tratados como resíduos domésticos comuns e devem ser eliminados deacordo com os regulamentos locais aplicáveis.
Quando utilizar o adaptador de CA, assegure que o adaptador de CA
pode ser prontamente retirado da tomada elétrica quando necessário.
Não modifique o dispositivo. A eventual modificação do dispositivo podeprovocar acidentes ou danos neste.
Para medir a tensão arterial, deve apertar o braço com a braçadeira osuficiente para deter temporariamente o fluxo sanguíneo através da
artéria. Tal pode provocar dor, dormência ou uma marca vermelha
temporária no braço. Este estado aparece especialmente quando amedição se repete sucessivamente. Qualquer dor, dormência ou marcas
vermelhas desaparecem com o tempo.
Se medir a tensão arterial com demasiada frequência, tal pode resultar
em danos devido à interferência com o fluxo sanguíneo. Se utilizar odispositivo de forma repetida, verifique se a utilização deste não resulta
numa disfunção prolongada da circulação sanguínea.
Caso tenha sido submetido a uma mastectomia, consulte o seu médico
antes de utilizar o dispositivo.
Não deixe que as crianças utilizem o dispositivo sem a supervisão de um
adulto, nem utilize o dispositivo num local dentro do alcance de crianças.
Tal pode provocar acidentes ou danos.
O dispositivo contém peças pequenas que podem provocar asfixia caso
as crianças as ingiram acidentalmente.
Retire o adaptador de CA do dispositivo quando não estiver a utilizá-lo
durante a medição.
O uso de acessórios não indicados neste manual pode comprometer asegurança.
Caso as pilhas entrem em curto-circuito, estas podem ficar quentes e
eventualmente causar queimaduras.
Permita que o dispositivo se adapte ao ambiente circundante antes deutilizá-lo (cerca de uma hora).
Não foram realizados ensaios clínicos em recém-nascidos e em mulheres
grávidas. Não usar em recém-nascidos ou em mulheres grávidas.
Não toque simultaneamente nas pilhas, na tomada de CC e no doente.
Tal pode resultar numa descarga elétrica.
Não encha sem colocar a braçadeira no antebraço.

Português 3

Contraindicações

Em seguida, indicam-se as precauções para o uso correto do dispositivo.
Não aplique a braçadeira num braço com outro dispositivo elétrico
médico conectado. O equipamento pode não funcionar corretamente.
As pessoas com um grave défice circulatório no braço devem consultar
um médico antes de utilizar o dispositivo, de modo a evitar problemasmédicos.
Não autodiagnostique os resultados de medição nem inicie o tratamento
por iniciativa própria. Consulte sempre o seu médico para a avaliação
dos resultados e tratamento.
Não aplique a braçadeira num braço com uma ferida por curar.
Não aplique a braçadeira num braço que está a receber tratamento por
via intravenosa ou transfusão sanguínea. Tal pode provocar lesões ou
acidentes.
Não utilize o dispositivo na presença de gases inflamáveis como, por
exemplo, gases anestésicos. Tal pode provocar uma explosão.
Não utilize o dispositivo em ambientes com alta concentração deoxigénio, tais como uma câmara de oxigénio de alta pressão ou uma
tenda de oxigénio. Tal pode provocar um incêndio ou explosão.

Português 4

Identificação de componentes
Ecrã

Ficha do conetor de ar

Ficha adaptadora de CA

Tomada de CCBotão START
Bocal de ar

Gama adequada deajusteTubo de arMarca de posição da artéria

Marca de índice
BraçadeiraCompartimento das pilhas
Tampa das pilhas

Pilhas de 1,5 V(R6P, LR6 ou AA)

Ecrã

Média
MEMÓRIA
Símbolo de IHB/AFib
Sistólica
Pressão

%IHB/AFib
Símbolo de erro de ajuste da braçadeira

Símbolo de erro por movimentoTensão diastólica

Marca de coração
Frequência depulsação

Indicador de classificação segundo a OMS
e indicador da barra de tensão

Indicação de pilhas
Português 5

Símbolos
Símbolos impressos na caixa do dispositivo
Símbolos
SYS
DIA
PUL

IP

Função/significado
Dispositivo em modo de espera e ligar o dispositivo.
Tensão arterial sistólica em mmHg
Tensão arterial diastólica em mmHg
Pulsação por minuto
Guia de instalação das pilhas
Corrente contínua
Tipo BF: o dispositivo, a braçadeira e o tubo foram desenvolvidospara proporcionar proteção especial contra choques elétricos.
Etiqueta de dispositivos médicos segundo diretiva CE
Representante da UE
FabricanteData de fabrico
Símbolo de proteção internacional
Etiqueta REEE

SN

Número de série
Consulte o manual/folheto de instruções
Polaridade da tomada de CC
Manter seco

Símbolos que são apresentados no ecrãSímbolosFunção/significadoAção recomendada
É apresentado quando a medição está A medição está emem curso. Este indicador pisca quando curso. Permanecer oa pulsação é detetada.
mais imóvel possível.
Símbolo de IHB/AFib
É apresentado quando se deteta um
batimento cardíaco irregular.
Pode acender quando se deteta uma
vibração muito ligeira como tremor ou
agitação.
A leitura pode produzir
um valor incorreto.
Aparece quando se deteta um
Realize a medição
movimento do corpo ou do braço.
novamente. Permaneça
imóvel durante a medição.
Português 6

Símbolos
SímbolosFunção/significadoAção recomendada

É apresentado durante a medição
quando a braçadeira não está bem
ajustada

A leitura pode produzir um
valor incorreto. Aplique a
braçadeira corretamente e
realize a medição
novamente.

Índice detetado de IHB/AFib na
memória%IHB/AFib =Número de IHB/AFibs
detetados na memóriax100 [%]

Número total

Medições anteriores armazenadas naMEMÓRIA.
Dados médios
PILHAS CARREGADAS
Indicador da potência das pilhas
durante a medição.
PILHA FRACAAs pilhas estão fracas quando este
símbolo pisca.
Tensão arterial instável devido amovimento durante a medição.

Substitua todas as pilhas
por novas quando a marca
piscar.
Realize a medição
novamente.
Permaneça totalmente
imóvel durante a medição.

Os valores sistólicos e diastólicos têm
um intervalo máximo de 10 mmHg
entre si.
O valor da tensão não aumentou
durante a inflação.
A braçadeira não está colocada
corretamente.
ERRO DO ECRÃ PUL
A pulsação não é detetada
corretamente.

Aplique a braçadeiracorretamente e realize amedição novamente.

Erro interno do dispositivo demonitorização de tensão arterial

Remova as pilhas,pressione o botão STARTe, em seguida, volte a
colocar as pilhas. Se oerro continuar a aparecer,
contacte o seu distribuidor.

Português 7

Modo de funcionamento
1. Medição normal
Pressione o botão START . É medida a tensão arterial e os dados são
armazenados na memória. Este dispositivo pode armazenar as últimas 60
medições na memória.

2. Revisão dos dados da memória
Quando não visualizar quaisquer dados,pressione e mantenha o botão START
pressionado.

Em espera,
mantenhapressionado o
START

botão

Solte o botão quando visualizar os dados
médios.

O número de dados e os dados armazenados
são visualizados automaticamente por ordem
desde a última medição.

Para mais informações sobre a revisão dedados, consulte “Revisão dos dados damemória”.

3. Eliminação de todos os dados armazenados na memória
Quando desligar o dispositivo, mantenha pressionado”.
o botão START até visualizar “Selecione “” para apagar os dados.
Os dados são apagados quando a marca
piscar.
O dispositivo desliga automaticamente.

START

4. Medição com a tensão sistólica desejada
Consulte a página 15 para a medição com a tensão sistólica desejada.
Português 8

Utilização do dispositivo demonitorização
Instalação/substituição das pilhas
1. Retire a tampa das pilhas.
2. Retire as pilhas usadas e insira aspilhas novas no compartimento daspilhas,tal
comodescrito,
respeitando as polaridades (+ e -)das pilhas.
Utilize apenas pilhas R6P, LR6 ou
AA.
3. Coloque a tampa das pilhas.

Passo 1
Passo 3

Passo 1

Passo 2CUIDADOInsira as pilhas no compartimento das pilhas, tal como ilustrado na figura.
Caso contrário, o dispositivo não funcionará.
Quando(marca de PILHA FRACA) piscar no ecrã, substitua todas aspilhas por pilhas novas. Não misture pilhas usadas com pilhas novas. Talpode encurtar a vida útil das pilhas ou causar o mau funcionamento do
dispositivo.
Substitua as pilhas pelo menos dois segundos após o dispositivo desligar.
Se(marca de PILHA FRACA) aparecer até mesmo depois desubstituir as pilhas, realize uma medição da tensão arterial. O dispositivo
pode então reconhecer as pilhas novas.
(marca de PILHA FRACA) não aparece quando as pilhas estão
totalmente descarregadas.
A vida útil das pilhas varia com a temperatura ambiente e pode serreduzida com temperaturas baixas. Em geral, quatro pilhas LR6 novas
duram aproximadamente um ano quando utilizadas para realizar medições
duas vezes por dia.
Utilize apenas as pilhas especificadas. As pilhas fornecidas com odispositivo são para testar o funcionamento do dispositivo demonitorização e podem ter uma vida útil mais curta.
Retire as pilhas quando não pretender utilizar o dispositivo durante um
período prolongado.
As pilhas podem ter fugas e causar um mau funcionamento.

Português 9

Utilização do dispositivo demonitorização
Ligação do tubo de ar

Bocal de arb
ldInsira firmemente a ficha doconetor de ar no bocal de ar.

Ficha do conetor de ar

Ligação do adaptador de CA
Insira a ficha do adaptador de CA
na tomada de CC.
Em seguida, ligue o adaptador deCA à tomada elétrica.
Utilize o adaptador de CA especificado.
(Consulte a página 22.)

Tomada de CC
Ficha adaptadora de CA

Quando desligar o adaptador de CA da tomada elétrica, agarre e puxe
o corpo do adaptador de CA para fora da tomada.
Quando desligar a ficha adaptadora de CA do dispositivo demonitorização de tensão arterial, agarre e puxe a ficha adaptadora de
CA para fora do dispositivo de monitorização.

Português 10

Utilização do dispositivo demonitorização
Seleção do tamanho correto da braçadeira
A seleção de uma braçadeira de tamanho correto é importante para uma leitura
correta. Se a braçadeira não for do tamanho correto, a leitura pode indicar um
valor incorreto da tensão arterial.
O tamanho do braço está impresso em cada braçadeira.
O índicee a gama adequada de ajuste da braçadeira indicam-lhe se está acolocar a braçadeira corretamente. (Consulte os “Símbolos impressos na
braçadeira” na página seguinte.)
Se o índice
se encontrar fora da gama, entre em contacto com o distribuidor
local para adquirir uma braçadeira de substituição.
A braçadeira é um produto consumível. Caso esta se desgaste, adquira uma nova.
Tamanho da braçadeira
Número de catálogo
recomendado
31 cm a 45 cm
Braçadeira grande para adultos
CUF-F-LA
22 cm a 42 cm
Braçadeira de grande extensão
CUF-I
22 cm a 32 cm
Braçadeira para adultos
CUF-F-A
Tamanho do braço: a circunferência no bíceps.
Nota:o UA-767S não foi desenvolvido para utilização de uma
braçadeira pequena.
Tamanho do braço

Aplicação da braçadeira
1. Coloque a braçadeira em redor do
antebraço, cerca de 1 a 2 cm acima daparte interior do cotovelo, tal como
ilustrado na figura.
Coloque a braçadeira em contacto
direto com a pele, dado que a roupa
pode causar uma pulsação fraca e
provocar erro de medição.

Evite deixar a manga demasiado apertada
ao enrolá-la para cima.

Braçadeira
1 a 2 cm

Tubo de arMarca de posição da artéria

2. A constrição do antebraço, causada
ao enrolar a manga apertada, pode
impedir leituras precisas.

Aperte firmemente
o fixador de tecido.

se encontra
3. Confirme se o índicedentro da gama adequada de ajuste.
ÍndicePortuguês 11

e gama adequada de ajuste

Utilização do dispositivo demonitorização
Símbolos impressos na braçadeira
SímbolosFunção/significadoAção recomendadana artéria
Ajuste a marca
do antebraço ou de forma
alinhada com o dedo anelar
na parte interior do braço.

Marca de posição da artéria▲
REFÍndiceNúmero de catálogo
Gama adequada de ajuste dabraçadeira para adultos.
Está impressa na braçadeira para
adultos.
Utilize a braçadeira grande
Acima da gama impressa na para adultos em vez dabraçadeira para adultos e braçadeira braçadeira para adultos ou a
braçadeiradegrandede grande extensão.
extensão.
Gama adequada de ajuste da braçadeira
de grande extensão. Está impressa na
braçadeira de grande extensão.
Gama adequada de ajuste da braçadeira
grande para adultos. Está impressa na
braçadeira grande para adultos.
Abaixo da gama impressa na
braçadeira para adultos e braçadeira
de grande extensão.
Utilize a braçadeira para
Abaixo da gama na braçadeira grande
adultos em vez da braçadeirapara adultos.
grande para adultos.
Número de loteBraçadeira grande para adultos

Gama adequada de ajusteA

L

Braçadeira de grande extensão

S

W

L

A

L

Braçadeira para adultos

S

Português 12

Utilização do dispositivo demonitorização
Como efetuar medições precisas
Tenha em conta os seguintes aspetos para efetuar a medição da tensão
arterial com a máxima precisão:Sente-se comodamente numa cadeira. Apoie o seu braço em cima da
mesa. Não cruze as pernas. Mantenha os pés assentes no chão e endireite
as costas.
Relaxe durante cerca de cinco a dez minutos antes de efetuar a medição.
Coloque o centro da braçadeira ao mesmo nível que o seu coração.
Permaneça imóvel e em silêncio durante a medição.
Não efetue a medição imediatamente após praticar exercício físico ou tomar
banho. Descanse cerca de vinte ou trinta minutos antes de efetuar a medição.
Tente medir a sua tensão arterial à mesma hora todos os dias.

Medição
Durante a medição, é normal sentir a braçadeira muito apertada. (Não sealarme por isso.)

Após a medição
Após a medição, pressione o botão START para desligar o dispositivo.
Retire a braçadeira e registe os seus dados.
Nota: o dispositivo possui uma função de desativação automática daalimentação que desliga o dispositivo aproximadamente um minuto
após a medição.
Deixe decorrer pelo menos três minutos entre medições na mesma
pessoa.

Português 13

Medições
Antes de efetuar a medição, consulte “Notas para medições corretas” na página
seguinte.

Medição normal
1.

2.

Coloqueabraçadeiranobraço(preferencialmente no braço esquerdo).
Sente-se tranquilamente durante a medição.

Ao nível do coração

Pressione o botão START .
Pressione
São apresentados todos os segmentos doecrã.
É apresentado zero (0) a piscar brevemente.
São apresentados todos
A indicação no ecrã muda, conforme indicado
os segmentos do ecrãna figura à direita, no início da medição. A
braçadeira começa a encher. É normal sentir a
braçadeira muito apertada. É visualizado oindicador da barra de tensão durante a inflação
Ecrã zerono canto esquerdo do ecrã.
Inicia a inflação

START

Nota: se desejar parar a inflação a qualquer
altura, pressione novamente o botão
START .
3.

Quando a inflação estiver concluída, adeflação começa automaticamente e amarca(marca de coração) pisca,
indicando que a medição está em curso.
Uma vez detetada pulsação, a marca pisca
com cada batimento do coração.

Pressurização

Nota: se não obtiver uma tensão apropriada, odispositivo começa novamente a encher
automaticamente.
Para evitar uma nova inflação, consulte“Medição com a tensão sistólica desejada”
na página seguinte.
4.

5.

Quando a medição estiver concluída, sãovisualizadas as leituras das tensões
sistólica e diastólica e a frequência depulsação.
A braçadeira expulsa o ar residual e esvazia
completamente.
Pressione o botão
dispositivo.

START

A medição está
em cursoTensão sistólica
Tensão diastólica
Classificação segundo a OMS

para desligar o Frequência de pulsação

Nota: deixe decorrer pelo menos três minutos
entre medições na mesma pessoa.
Português 14

Esvazia automaticamenteo ar residual

Medições
Medição com a tensão sistólica desejada

O UA-767S foi desenvolvido para detetar a pulsação, enchendo
automaticamente a braçadeira até ao nível de tensão sistólica.
Utilize este método quando a repetição da inflação ocorrer ou quando os
resultados não forem visualizados, mesmo no caso em que a tensão
diminua para 20 mmHg ou menos.

1.

Coloque a braçadeira no braço ao nível docoração (preferencialmente no braçoesquerdo).

2.

Pressione o botão START .

3.

Quando o zero piscar, pressione e
mantenha o botão START
pressionado até aparecer um número
cerca de 30 a 40 mmHg superior à
sua tensão sistólica prevista.

4.

Quando o número desejado foratingido, solte o botão START para
iniciar a medição. Continue a medir asua tensão arterial conforme descritona página anterior.

Ao nível do coração
Prima o
botão

START

Solte o botão natensão sistólica
desejada

Pressione e
mantenha o botão
pressionado

START

Consulte a página anteriorpara a medição

Notas para medições corretas
Sente-se numa posição confortável. Coloque o braço em cima de uma mesa
com a palma virada para cima e a braçadeira ao mesmo nível que o coração.
Relaxe durante cerca de cinco a dez minutos antes de efetuar a medição.
Se estiver agitado ou deprimido devido a stress emocional, a medição irá
refletir este stress como uma leitura mais alta (ou mais baixa) do que aleitura da tensão arterial normal e a leitura da pulsação será geralmentemais rápida do que a normal.
A tensão arterial de uma pessoa varia constantemente, dependendo daquilo
que estiver a fazer e daquilo que comeu. Aquilo que bebe pode ter um efeito
muito forte e rápido sobre a sua tensão arterial.
Este dispositivo baseia as medições no batimento cardíaco. Se tiver um
batimento cardíaco muito fraco ou irregular, este dispositivo pode ter
dificuldade em determinar a sua tensão arterial.
No caso de o dispositivo detetar uma condição anormal, o mesmo
interrompe a medição e indica um símbolo de erro no ecrã. Consulte apágina 7 para a descrição dos símbolos.
Português 15

Este dispositivo de monitorização de tensão arterial destina-se à utilização em
adultos. Consulte o seu médico antes de utilizar este dispositivo numa criança.
Uma criança não deve utilizar este dispositivo sem a ajuda de um adulto.
O desempenho do dispositivo de monitorização de tensão arterial
automático pode ser afetado por temperatura ou humidade excessivas ou
pela altitude.

Revisão dos dados da memória
Nota: este dispositivo armazena as últimas 60 medições na memória.
1. Quando não forem visualizados
dados, pressione e mantenha obotão _START_ pressionado pararever os dados armazenados.

Pressione e
mantenha
START botão
pressionadoTensão sistólica médiaTensão diastólica média
Frequência média da
Média de pulsação
todos os dados

2. Solte o botão quando visualizar os
dados médios.

3. O número de dados e os dados
armazenados são visualizados
automaticamente por ordem desdea última medição.

Sistólica
Diastólica
Pulsação
Dados mais recentes

4. O ecrã desliga automaticamente
após todos os dados serem
visualizados.

Nota: Se pressionar o botão _START_
enquanto revê os dados, odispositivo desliga.
Sistólica
Diastólica
Pulsação
Últimos dados (mais antigos)
Português 16

O que é o indicador de IHB/AFib?
Se o dispositivo de monitorização detetar um ritmo irregular durante as medições,será apresentado o indicador de IHB/AFib no ecrã com os valores de medição.
Nota: recomendamos que consulte o seu médico se visualizar este
indicador de IHB/AFib com frequência.

A que se refere AFib?
O coração contrai devido a correntes elétricas que ocorrem no coração e envia
o sangue por todo o corpo. A fibrilhação auricular (AFib) ocorre quando ascorrente elétrica do átrio cardíaco se confundem e originam alterações nos
intervalos de pulsação. A AFib pode ocasionar estagnações de sangue nocoração, as quais podem produzir facilmente coágulos de sangue, uma dascausas de AVC e de enfartes cardíacos.

％IHB/AFib%IHB/AFib é visualizado como frequência de IHB/AFib detetado.
%IHB/AFib pode detetar não só interferências tais como o movimento físico, mastambém um batimento cardíaco irregular. Como tal, recomendamos que consulteo seu médico se o nível de %IHB/AFib for elevado.
%IHB/AFib =Número de IHB/FibsA
detetados na memória

x 100 [%]

Número totalEcrã de %IHB/AFib:

Nível 0
%IHB/AFib=0%IHB/AFib é visualizado quando são visualizados osvalores médios. (Consulte “2. Revisão dos dados damemória” em “Modo de funcionamento”)%IHB/AFib não é visualizado quando o número dememória for igual a seis ou inferior.
Visualização do valor médio

Nível 1
Nível 2
Nível 3
%IHB/AFib=1 - 9 %IHB/AFib=10 - 24 %IHB/AFib=25 - 100

Não visualizado

Português 17

Indicador da barra de tensão
O indicador monitoriza o progresso da tensão durante a medição.
Pressurização

Libertação do ar

Inflação em curso

Medição em curso
Inflação completa

Indicador de classificação
segundo a OMS
Cada um dos segmentos do indicador de barra
corresponde à classificação da tensão arterial
segundo a OMS descrita na página seguinte.

Exemplo:

Indicador de classificação segundo a OMS

Hipertensão grave
Hipertensão moderada
Hipertensão ligeira
Alta normal
Normal
Ideal

Hipertensão moderada

Hipertensão ligeira: O indicador visualiza um segmento baseado
nos dados atuais, correspondentes àclassificação segundo a OMS.

Alta normal

Sobre tensão arterial
O que é a tensão arterial?
A tensão arterial é a força que o sangue exerce contra as paredes dasartérias. A tensão sistólica ocorre quando o coração contrai. A tensão
diastólica ocorre quando o coração expande. A tensão arterial é medida em
milímetros de mercúrio (mmHg). A tensão arterial natural de uma pessoa é
representada pela tensão fundamental, a qual se mede logo pela manhã
quando a pessoa ainda está em repouso e antes de se alimentar.
Português 18

O que é a hipertensão e como se controla?
A hipertensão é a tensão arterial anormalmente alta e, caso não seja corrigida,pode causar muitos problemas de saúde, incluindo AVC e enfarte cardíaco. A
hipertensão pode ser controlada através da alteração do estilo de vida,
evitando o stress e com medicamentos mediante a supervisão de um médico.
Para evitar a hipertensão ou para mantê-la sob controlo:Não fume
Faça exercício regularmente
Reduza a ingestão de sal e de gordura
Faça exames físicos regulares
Mantenha o peso ideal

Porque deve medir a tensão arterial em casa?
A medição da tensão arterial numa clínica ou num consultório médico podecausar apreensão e, por isso, produzir uma leitura elevada, 25 a 30 mmHg
superior à medida em casa. A medição em casa reduz os efeitos de influências
externas sobre as leituras da tensão arterial, complementa as leituras do
médico e proporciona um historial mais completo e exato da tensão arterial.

Classificação da tensão arterial segundo a OMS

Variações da tensão
arterialTensão arterial diastólica

A Organização Mundial deMaterial de referência: Journal of Hypertension 1999, Vol. 17, N.º 2
mmHg
Saúde (OMS) estabeleceu
Hipertensão de grau 3 (grave)110
as normas para a avaliação
Hipertensão de grau 2 (moderada)
105
da tensão arterial,
100
independentemente da idade,Hipertensão de grau 1 (leve)95
conforme ilustrado no90
gráfico.
Alta-normal
85
80

Normal
Ideal
120

130

140 150 160 170Tensão arterial sistólica

180
mmHg

Tensão arterial

A tensão arterial de uma
pessoa varia consideravelmente numa base diária e sazonal. Esta pode variar
entre 30 a 50 mmHg dependendo de diversas condições durante o dia. Nas
pessoas hipertensas, estas variações são ainda mais notáveis. Normalmente,a tensão arterial sobe quando a pessoa está a trabalhar ou a jogar e baixa
para os seus níveis mais baixos durante o sono. Portanto, não se preocupe
demasiado com os resultados de uma única
Variação típica num dia
medição.
(Medição a cada cinco minutos)
Efetue as medições à mesma hora todos os
Sistólicadias, utilizando o procedimento descrito
neste manual e familiarize-se com a suatensão arterial normal. As leituras regulares
proporcionam um historial da tensão arterial
mais completo. Assegure que anota a data e
DormirDiastólica
a hora ao registar a sua tensão arterial.
Consulte o seu médico para interpretar os
Hora
PM
AM
dados da sua tensão arterial.

Português 19

Resolução de problemas
ProblemaNão são
visualizados
quaisquer
dados no ecrã,mesmo quando
o dispositivo
está ligado.
A braçadeiranão enche.

Causa provável
As pilhas estão gastas.
Os terminais das pilhasnão estão na posição
correta.
A tensão das pilhas é
demasiado baixa.
(marca de PILHAFRACA) pisca. Se as
pilhas estiverem
completamente gastas,a marca não aparece.
A braçadeira não está
colocada
adequadamente.
Deslocou o seu braço
ou corpo durante a
medição.

Ação recomendada
Substitua todas as pilhas por
novas.
Volte a colocar as pilhas com os
terminais negativos e positivos
em conformidade com os
terminais indicados nocompartimento das pilhas.
Substitua todas as pilhas por
novas.

Aplique a braçadeiracorretamente.

Assegure que permanece
imóvel e calmo durante a
O dispositivo
medição.
não efetua
Sente-secomodamente emedições.
imóvel.
As leituras sãoA
posição
dabraçadeira
Coloque o braço em cima dedemasiado
não é a correta.
uma mesa com a palma virada
altas oupara cima e a braçadeira ao
demasiado
mesmo nível que o coração.
baixas.
Se tiver um batimento cardíaco
muito fraco ou irregular, odispositivo pode ter dificuldade
em determinar a sua tensão
arterial.
O valor é diferente dovalor medido numa
Consulte “Porque deve medir aclínica ou numtensão arterial em casa?”.
consultório médico.
Outro
Retire as pilhas. Volte a
colocá-las corretamente e tente
efetuar a medição novamente.
Nota: se as medidas descritas anteriormente não solucionarem o problema,contacte o distribuidor. Não tente abrir ou reparar este produto, dado
que qualquer tentativa nesse sentido invalida a sua garantia.

Português 20

Manutenção
Não abra o dispositivo. Este contém componentes elétricos delicados e uma
unidade de ar complexa que podem sofrer danos. Caso não solucione oproblema através das instruções referentes à resolução de problemas, contacte
o distribuidor autorizado na sua área de residência ou o departamento de apoio
ao cliente da A&D. O serviço de apoio ao cliente da A&D fornece informações
técnicas, peças sobresselentes e unidades aos distribuidores autorizados.
O dispositivo foi desenvolvido e fabricado para atingir uma vida útil prolongada.
Não obstante, recomenda-se geralmente que o dispositivo seja submetido a
uma inspeção a cada 2 anos a fim de assegurar o funcionamento e a precisão
adequados. Contacte o distribuidor autorizado na sua área de residência ou a
A&D para efeitos de manutenção.

Dados técnicos
Tipo
Método de medição
Intervalo de medição

UA-767S
Medição oscilométricaTensão:0 a 299 mmHg
Tensão sistólica: 60 a 279 mmHg
Tensão diastólica: 40 - 200 mmHg
Pulsação: 40 a 180 batimentos/minuto
Intervalo de medição
Tensão:±3 mmHg
Pulsação: ±5%
Fonte de alimentação
4 x pilhas de 1,5 V (R6P, LR6 ou AA) ou
adaptador de CA (TB-233C) (não incluído)
Número de medições
Aprox. 700 vezes
LR6 (pilhas alcalinas)
Aprox. 200 vezes
R6P (pilhas de manganês)Com valor da tensão de 180 mmHg a uma
temperatura ambiente de 23 °C.
Classificação
Equipamento ME alimentado internamente(alimentado por pilhas)/Classe II (alimentado por
adaptador)
Modo de funcionamento contínuo
Ensaio clínico
De acordo com a ISO81060-2：2013
No estudo de validação clínica, foi utilizado K5 em 85
doentes para determinação da tensão arterial diastólica.
EMD
IEC 60601-1-2: 2014
MemóriaÚltimas 60 medições
Condições de funcionamento +10 a +40 °C/15 a 85% de RH/800 a 1.060 hPa
Condições de transporte/-20 a +60 °C/10 a 95% de RH/700 a 1.060 hPa
armazenamento
Dimensões
Aprox. 140 [L] x 60 [A] x 105 [P] mm
Peso
Aprox. 245 g, excluindo as pilhas
Proteção
Dispositivo: IP20
Português 21

Componente aplicado
Vida útil

Braçadeira Tipo BF
Dispositivo: 5 anos (quando utilizado seis vezes por dia)
Braçadeira: 2 anos (quando utilizada seis vezes por dia)
Adaptador de CA: 5 anos (quando utilizado seis vezes
por dia)

Adaptador de CA auxiliar
O adaptador serve para ligar o dispositivo a uma fonte de alimentação em casa.
Contacte o seu distribuidor A&D local para efetuar a aquisição. O adaptador de
CA deve ser inspecionado ou substituído periodicamente.
TB-233C
Contacte o seu distribuidor A&D local para efetuar aaquisição.
O adaptador de CA deve ser inspecionado ou
substituído periodicamente.
Símbolos impressos no adaptador de CA
SímbolosFunção/significado
Exclusivamente para uso interno
Dispositivo de classe II
Fusível térmico
FusívelEtiqueta de dispositivos segundo diretivade CE
Etiqueta de dispositivos segundo acertificação EAC
Polaridade da ficha adaptadora de CA
Acessórios vendidos separadamente
Braçadeira Número de catálogo Tamanho da braçadeira Tamanho do braço
Braçadeira grande para
CUF-F-LA
31 cm a 45 cm
adultosBraçadeira de grande
CUF-I
22 cm a 42 cm
extensão
CUF-F-A
Braçadeira para adultos
22 cm a 32 cm
Tamanho do braço: A circunferência no bíceps.
Adaptador de CA
Nota:

Número de catálogo
TB-233C

Conetor (tipo de tomada)
Tipo C

as especificações estão sujeitas a alteração sem aviso prévio.
A classificação IP corresponde aos graus de proteção fornecidos pelas
caixas de acordo com a IEC 60529. Este dispositivo está protegido contra
objetos estranhos sólidos com 12 mm de diâmetro e maiores do que os
dedos, por exemplo. Este dispositivo não está protegido contra água.

Português 22

Inhoud
Beste klant ............................................................................ 2
Inleidende opmerkingen ....................................................... 2
Voorzorgsmaatregelen.......................................................... 2
Identificatie onderdelen ........................................................ 5
Symbolen.............................................................................. 6
Gebruiksmodus .................................................................... 8
De bloeddrukmonitor gebruiken ........................................... 9
De batterijen plaatsen / vervangen .................................. 9
De luchtslang aansluiten ................................................ 10
De netstroomadapter aansluiten .................................... 10
De juiste manchetgrootte selecteren .............................. 11
De armmanchet aanbrengen .......................................... 11
Nauwkeurige metingen uitvoeren ................................... 13
Meting ........................................................................... 13
Na meting ...................................................................... 13
Metingen ............................................................................ 14
Normale meting ............................................................. 14
Meting met de gewenste systolische druk ...................... 15
Een nauwkeurige meting uitvoeren ................................ 15
De gegevens uit het geheugen opvragen............................. 16
Waarvoor dient de IHB/AFib-indicator? ............................... 17
Wat is de AFib? ................................................................... 17％IHB/AFib .......................................................................... 17
Drukbalkindicator ............................................................... 18
Indicator WHO-classificatie ................................................. 18
Meer over bloeddruk........................................................... 18
Wat is bloeddruk? .......................................................... 18
Wat is hoge bloeddruk en hoe wordt deze onder
controle gehouden? ....................................................... 19
Waarom thuis bloeddruk meten? .................................... 19
Bloeddrukclassificatie WHO ............................................ 19
Bloeddrukvariaties ......................................................... 19
Probleemoplossing ............................................................. 20
Onderhoud ......................................................................... 21
Technische gegevens .......................................................... 21

Nederlands 1

Beste klant
Gefeliciteerd met uw aanschaf van een hypermoderne A&D-bloeddrukmonitor.
Dit apparaat is ontworpen voor gebruiksgemak en nauwkeurigheid en zal uw
dagelijkse bloeddrukmetingen vergemakkelijken.
We bevelen u aan deze handleiding zorgvuldig door te lezen voordat u het
apparaat voor het eerst gebruikt.

Inleidende opmerkingenDit apparaat voldoet aan Europese Richtlijn 93/42/EEG betreffende medische.
hulpmiddelen. Dit wordt aangegeven met de conformiteitsmarkering(0123: het referentienummer voor de betrokken aangemelde instantie)
Het apparaat is ontworpen voor gebruik bij volwassenen, niet bij baby’s of peuters.
Gebruiksomgeving. Het apparaat is bedoeld voor gebruik door uzelf voor
medische zorg in de thuissituatie.
Dit apparaat is ontworpen voor het meten van bloeddruk en hartslag vanpersonen voor diagnosedoeleinden.

Voorzorgsmaatregelen
Voor de constructie van dit apparaat zijn precisieonderdelen gebruikt. Vermijd
extreme temperatuur, luchtvochtigheid, direct zonlicht, schokken of stof.
Reinig het apparaat en de manchet met een droge, zachte doek of een doek
die is bevochtigd met water en een neutraal schoonmaakmiddel. Gebruik in
geen geval alcohol, benzeen, thinner of andere agressieve chemische
middelen om het apparaat en de manchet te reinigen.
Vouw de manchet niet te strak op en bewaar de slang niet gedurende langere
perioden in strak opgewikkelde vorm, aangezien dit de levensduur van deonderdelen kan verkorten.
Pas op dat baby’s of peuters zichzelf niet per ongeluk wurgen met de slang en
kabel.
Draai de luchtslang niet tijdens het meten. Dit zou letsel kunnen veroorzaken
als gevolg van continue druk van de manchet.
Het apparaat en de manchet zijn niet waterbestendig. Voorkom dat het
apparaat en de manchet in contact komen met regen, zweet en water.
Metingen kunnen verkeerde waarden aangeven als het apparaat wordt
gebruikt in de nabijheid van televisies, magnetrons, mobiele telefoons,röntgenapparaten of andere apparaten met een sterk elektrisch veld.
Draadloze communicatieapparaten zoals thuisnetwerkapparaten, mobiele
telefoons, draadloze telefoons en de basisstations daarvan en walkietalkies
kunnen van invloed zijn op de bloeddrukmonitor. Hanteer daarom een
minimumafstand van 30 cm van dergelijke apparaten.
Nederlands 2

Wanneer u het apparaat opnieuw gebruikt, dient u eerst te controleren of het
schoon is.
Gebruikte apparatuur, onderdelen en batterijen mogen niet als normaal
huishoudelijk afval worden behandeld en moeten worden afgevoerd volgens detoepasselijke plaatselijke regels.
Wanneer u de netstroomadapter gebruikt, dient u ervoor te zorgen dat deadapter snel uit het stopcontact kan worden gehaald indien nodig.
Breng geen wijzigingen aan in het apparaat. Dit kan ongevallen of schade aan
het apparaat veroorzaken.
Als u de bloeddruk wilt meten, moet de arm hard genoeg worden afgekneld
door de manchet dat de bloedstroom door de slagader tijdelijk wordt gestopt.
Dit kan pijn, gevoelloosheid of een tijdelijke rode verkleuring van de arm
opleveren. Dit doet zich met name voor wanneer de meting meerdere keren inkorte tijd wordt herhaald. Eventuele pijn, gevoelloosheid of rode verkleuringen
verdwijnen na verloop van tijd.
Het te vaak meten van de bloeddruk kan kwalijk zijn doordat de bloedcirculatie
wordt verstoord. Controleer dat de werking van het apparaat niet resulteert inlangdurige verstoring van de bloedcirculatie wanneer u het apparaat
herhaaldelijk gebruikt.
Als u een borstamputatie hebt ondergaan, dient u een arts te raadplegenvoordat u het apparaat gebruikt.
Zorg ervoor dat kinderen het apparaat niet zonder toezicht gebruiken en
gebruik het apparaat niet binnen bereik van baby’s of peuters. Dit kan
ongevallen of schade veroorzaken.
Het apparaat bevat kleine onderdelen die een risico op verstikking opleveren
indien ze door baby’s of peuters worden ingeslikt.
Koppel de netstroomadapter los wanneer deze niet in gebruik is tijdens het meten.
Het gebruik van accessoires die niet in deze handleiding worden vermeld, kan
een veiligheidsrisico opleveren.
Indien de batterij kortsluiting maakt, kan deze heet worden, met mogelijk
brandwonden tot gevolg.
Laat het apparaat (ongeveer een uur) op kamertemperatuur komen voordat u
het gebruikt.
Er zijn geen klinische tests uitgevoerd bij pasgeborenen en zwangere vrouwen.
Gebruik het apparaat niet bij pasgeborenen of zwangere vrouwen.
Zorg ervoor dat u de batterijen, de DC-ingang en de patiënt niet tegelijkertijd
aanraakt. Dit zou een elektrische schok kunnen veroorzaken.
Pomp de manchet niet op voordat u deze rond de bovenarm hebt gewikkeld.

Contra-indicaties

Hierna volgen voorzorgsmaatregelen voor het juiste gebruik van het apparaat.
Breng de manchet niet aan rond een arm waar ook andere medischeelektronica zijn aangesloten. Het apparaat werkt dan mogelijk niet goed.

Nederlands 3

Mensen die ernstige circulatieproblemen hebben in de arm, moeten eerst een
arts raadplegen voordat ze het apparaat gebruiken. Dit om medische
problemen te voorkomen.
Voer geen zelfdiagnose uit aan de hand van de meetwaarden en begin niet zelf
een behandeling. Raadpleeg altijd eerst uw arts voor beoordeling en behandeling.
Breng de manchet niet aan rond een arm met een nog niet genezen wond.
Breng de manchet niet aan rond een arm die voor een infuus of bloedtransfusie
wordt gebruikt. Dit kan letsel of ongevallen veroorzaken.
Gebruik het apparaat niet op een plaats waar brandbare gassen aanwezig zijn,
zoals narcosegassen. Dit zou een explosie kunnen veroorzaken.
Gebruik het apparaat niet in omgevingen met een hoge zuurstofconcentratie,
zoals een zuurstofkamer met hoge druk of een zuurstoftent. Dit kan brand of
een explosie veroorzaken.

Nederlands 4

Identificatie onderdelen
Display

Connector
netstroomadapter

Connector luchtaansluiting

DC-ingang
START-knop
Aansluitpunt lucht

Luchtslang

Het juiste maatbereik

Slagadermarkering
Indexmarkering
Armmanchet
Batterijvak
Batterijklep

1,5V-batterijen(R6P, LR6 of AA)

Display

GemiddeldeGEHEUGENIHB/AFib-symbool
Systolischdruk%IHB/AFib
Symbool onjuist aangebrachte manchet

Diastolische
druk

Symbool bewegingsfout
Hartsymbool

HartslagIndicator WHO-classificatie
en drukbalkindicator
Batterij-indicator

Nederlands 5

Symbolen
Symbolen die op de behuizing van het apparaat zijn gedrukt
Symbolen
SYS
DIA
PUL

Functie / betekenis
Stand-by en Apparaat aanzetten.
Systolische bloeddruk in mmHg
Diastolische bloeddruk in mmHg
Hartslag per minuut
Installatie-instructies batterij
Gelijkstroom (DC)Type BF: apparaat, manchet en slangen zijn ontworpen om specialebescherming te bieden tegen elektrische schokken.
Label EG-richtlijn medische apparaten
EU-vertegenwoordiger
Fabrikant
Productiedatum

IP

Internationaal beveiligingssymbool
WEEE-label

SN

Serienummer
Zie de instructiehandleiding/het instructieboekje
Polariteit van DC-ingang
Droog houden

Symbolen die in de display worden weergegeven
Symbolen

Functie / betekenis

Aanbevolen handeling

Wordt weergegeven wanneer een meting De meting wordt
uitgevoerd. Blijf zo stilwordt uitgevoerd. Het knippert wanneer
mogelijk zitten.
de hartslag wordt waargenomen.
IHB/AFib-symbool
Wordt weergegeven wanneer een
onregelmatige hartslag wordt
waargenomen.
Het kan gaan branden wanneer een zeer
lichte vibratie, zoals een rilling ofsiddering, wordt waargenomen.
Er kan een onjuiste
meetwaarde worden
Wordt weergegeven wanneer een
weergegeven.
beweging van lichaam of arm wordt
Voer de meting opnieuw
waargenomen.
uit. Blijf stil zitten tijdens
het meten.
Nederlands 6

Symbolen
Symbolen

Functie / betekenis

Aanbevolen handeling

Er kan een onjuiste
meetwaarde worden
Wordt tijdens meting weergegeven als
weergegeven. Breng dede manchet te los zit
manchet correct aan envoer de meting opnieuw uit.
Gedetecteerd percentage IHB/AFib in
geheugen%IHB/AFib =Aantal gedetecteerde
IHB/AFib-voorvallen in
geheugen
Totaal aantal

x 100 [%]

Eerdere metingen die in GEHEUGEN
zijn opgeslagen.
Gemiddelde waarden
BATTERIJ VOL
De batterijstroomindicator tijdens
meting.
BATTERIJ BIJNA LEEG
Als dit symbool knippert, is de batterij
bijna leeg.
Onstabiele bloeddruk als gevolg van
beweging tijdens meting.

Vervang alle batterijen door
nieuwe exemplaren wanneer
dit symbool knippert.
Voer de meting opnieuw uit.
Blijf helemaal stil zitten
tijdens het meten.

De systolische en diastolische waarden
liggen binnen 10 mmHg van elkaar.
De drukwaarde is niet toegenomen
Breng de manchet correct
tijdens het opblazen.
aan en voer de meting
De manchet is niet correct
opnieuw uit.
aangebracht.
FOUT HARTSLAGWEERGAVE
De hartslag is niet goed gedetecteerd.
Verwijder de batterijen, druk
op de knop START en
plaats de batterijen terug.
Interne fout bloeddrukmonitor
Neem contact op met deverkoper als de fout nog
steeds wordt weergegeven.
Nederlands 7

Gebruiksmodus
1. Normale meting
Druk op de knop START . De bloeddruk wordt gemeten en de gegevens worden inhet geheugen opgeslagen. Dit apparaat kan de laatste 60 metingen in het
geheugen opslaan.

2. De gegevens opvragen

Houd de knop ingedrukt

Als niets wordt weergegeven, drukt u opnieuw op terwijl het apparaat
de knop START en houdt u deze ingedrukt.
START

knop

Laat de knop los wanneer de gemiddeldewaarden worden weergegeven.

Het gegevensnummer en de opgeslagengegevens worden automatisch op volgorde
weergegeven vanaf de laatste meting.

Raadpleeg “De gegevens uit het geheugen
opvragen“ voor meer informatie over het opvragenvan de gegevens.

3. Alle gegevens in het geheugen wissen
Terwijl u het apparaat uitzet, houdt u de knop
START ingedrukt tot de “
“ wordt weergegeven.
Selecteer ““ om de gegevens te wissen.
knippert.
De gegevens zijn gewist als het symbool
Het apparaat wordt automatisch uitgezet.

4. Meting met de gewenste systolische druk
Zie pagina 15 voor metingen met de gewenste systolische druk.

Nederlands 8START

De bloeddrukmonitor gebruiken
De batterijen plaatsen / vervangen
1. Verwijder de batterijklep.
2. Verwijder de gebruikte batterijen enplaats nieuwe batterijen in het
batterijvak, zoals hier weergegeven.
Zorg ervoor dat de polariteit (+ en -)correct is.
Gebruik uitsluitend R6P-, LR6- of
AA-batterijen.
3. Zet de batterijklep weer op zijn plaats.

Stap 1
Stap 3

Stap 1
Stap 2

LET OP
Plaats de batterijen in het batterijvak zoals weergegeven. Als ze niet goed
worden geplaatst, werkt het apparaat niet.
Als het symbool(BATTERIJ BIJNA LEEG) knippert op de display, dient u
alle batterijen door nieuwe te vervangen. Mix nooit oude en nieuwe batterijen. Dit
kan de levensduur van de batterijen verkorten. Daarnaast kan het ervoor zorgen
dat het apparaat defect raakt.
Vervang de batterijen twee of meer seconden nadat u het apparaat hebt uitgezet.
(BATTERIJ BIJNA LEEG) zelfs na vervanging van deAls het symbool
batterijen wordt weergegeven, dient u een bloeddrukmeting uit te voeren. Het ismogelijk dat het apparaat de nieuwe batterijen dan wel herkent.
Het symbool (BATTERIJ BIJNA LEEG) wordt niet weergegeven als debatterijen helemaal leeg zijn.
De levensduur van de batterijen varieert al naargelang de omgevingstemperatuur
en kan korter zijn bij lage temperaturen. Vier nieuwe LR6-batterijen gaan
doorgaans ongeveer een jaar mee wanneer u het apparaat tweemaal daags voor
metingen gebruikt.
Gebruik alleen de aangegeven batterijen. De batterijen die met het apparaat
worden meegeleverd, zijn bedoeld voor het testen van de bloeddrukmonitor enhebben mogelijk een beperkte levensduur.
Verwijder de batterijen als u het apparaat langere tijd niet gebruikt.
De batterijen kunnen gaan lekken en een defect veroorzaken.

Nederlands 9

De bloeddrukmonitor gebruiken
De luchtslang aansluiten
Plaats de connector van deluchtslang stevig in het aansluitpunt
voor lucht.

Aansluitpunt
lucht

Connector
luchtaansluiting

De netstroomadapter aansluiten
Plaats de connector voor denetstroomadapter in de DC-ingang.
Sluit de netstroomadapter vervolgens
aan op een stopcontact.
Gebruik de gespecificeerde netstroomadapter.
(Zie pagina 22.)

DC-ingang
Connector
netstroomadapter

Wanneer u de netstroomadapter wilt loskoppelen van het stopcontact, moetu dit doen door de adapter zelf vast te grijpen en deze uit het stopcontact tetrekken.
Wanneer u de netstroomadapter wilt loskoppelen van de bloeddrukmonitor,
moet u dit doen door de connector vast te grijpen en deze uit debloeddrukmonitor te trekken.

Nederlands 10

De bloeddrukmonitor gebruiken
De juiste manchetgrootte selecteren
Voor het verkrijgen van een nauwkeurige meting, is het belangrijk dat u de juiste
manchetgrootte gebruikt. Als de manchet niet de juiste maat heeft, kan de meting
een onjuiste bloeddrukwaarde opleveren.
De armomvang is op elke manchet gedrukt.
De index- en het juiste maatbereik op de manchet laten u zien of u de juiste
manchet gebruikt. (Zie “Symbolen die op de manchet zijn gedrukt” op devolgende pagina.)
Als de index- buiten het maatbereik wijst, dient u contact op te nemen met uw
plaatselijke verkoper om een vervangende manchet te kopen.
De armmanchet is een verbruiksitem. Koop een nieuw exemplaar als de oude
versleten raakt.
Armomvang
Aanbevolen manchetgrootte
Catalogusnummer
31 cm tot 45 cm
Grote manchet volwassenen
CUF-F-LA
22 cm tot 42 cm
Manchet voor breed spectrum
CUF-I
22 cm tot 32 cm
Manchet volwassenen
CUF-F-A
Armomvang: de omtrek van de arm ter hoogte van de biceps.
Opmerking: de UA-767S is niet ontworpen voor gebruik met een kleine manchet.

De armmanchet aanbrengen
1. Wikkel de manchet rond de bovenarm,ongeveer 1-2 cm boven de binnenzijde
van de elleboog, zoals in de illustratie.
Plaats de manchet direct tegen de huid.
Kleding kan een zwakkere
hartslagmeting veroorzaken en kan zo
een meetfout tot gevolg hebben.

Rol uw mouwen niet te strak op.
Manchet1-2 cmLuchtslang
Slagadermarkering

2. Het afklemmen van de bovenarm
door het te strak oprollen van een
mouw kan voor een onnauwkeurige
meting zorgen.
3. Ga na of de indexmaatbereik valt.

Maak de klittenbandsluiting stevig vast.

binnen het juiste
Index-

Nederlands 11

en het juiste maatbereikDe bloeddrukmonitor gebruiken
Symbolen die op de manchet zijn gedrukt
Symbolen Functie / betekenis

Aanbevolen handeling
Plaats de-markering op
deslagadervandebovenarm of in één lijn met
de ringvinger aan debinnenzijde van de arm.

Slagadermarkering
▲
REF

Index
Catalogusnummer
Het juiste maatbereik voor de manchet
voor volwassenen.
Het is gedrukt op de manchet voor
volwassenen.
Gebruik de grote manchet
Groter dan het bereik dat is gedrukt op voor volwassenen in plaatsde manchet voor volwassenen en de van de manchet voor
manchet voor breed spectrum.
volwassenen of de manchet
voor breed spectrum.
Het juiste maatbereik voor de manchet
voor breed spectrum. Het is gedrukt op
de manchet voor breed spectrum.
Het juiste maatbereik voor de grotemanchet voor volwassenen. Het isgedrukt op de grote manchet voor
volwassenen.
Kleiner dan het bereik dat is gedrukt op
de manchet voor volwassenen en demanchet voor breed spectrum.
Gebruik de manchet voor
Kleiner dan het bereik voor de grote volwassenen in plaats vanmanchet voor volwassenen.
de grote manchet voor
volwassenen.
Partijnummer

Grote manchet volwassenen

Het juiste maatbereik

A

L

Manchet voor breed spectrum

S

W

L

A

L

Manchet volwassenen

S

Nederlands 12

De bloeddrukmonitor gebruiken
Nauwkeurige metingen uitvoeren
Ga als volgt te werk voor de meest nauwkeurige bloeddrukmeting:
Ga comfortabel op een stoel zitten. Laat uw arm op de tafel rusten. Zorg ervoor
dat u uw benen niet kruist. Houd uw voeten plat op de vloer en houd uw rug recht.
Ontspan gedurende vijf tot tien minuten voordat u gaat meten.
Plaats het midden van de manchet ter hoogte van uw hart.
Blijf stil zitten en praat niet tijdens het meten.
Voer de meting niet uit als u zich net fysiek hebt ingespannen of een bad hebt
genomen. Rust ongeveer twintig tot dertig minuten voordat u de meting uitvoert.
Probeer uw bloeddruk iedere dag rond dezelfde tijd te meten.

MetingHet is normaal als de manchet erg knelt tijdens het meten. (Maak u niet ongerust.)

Na meting
Druk na meting op de knop START om het apparaat uit te zetten.
Verwijder de manchet en registreer uw gegevens.
Opmerking: het apparaat heeft een automatische uitschakelfunctie, waardoor het
apparaat ongeveer een minuut na de meting wordt uitgezet.
Gebruik een tussenpauze van minstens drie minuten voor metingen bij
dezelfde persoon.

Nederlands 13

Metingen
Lees voordat u gaat meten “Een nauwkeurige meting uitvoeren“ op devolgende pagina.

Normale meting
1.

Plaats de manchet rond de arm (bij voorkeur delinkerarm).
Ter hoogte van het hart
Zit stil tijdens het meten.

2.

Druk op de knop START .
Alle displayonderdelen worden weergegeven.
Nul (0) wordt weergegeven en knippert een
korte tijd.
De weergave op de display verandert, zoals
te zien op de afbeelding rechts, op het
moment dat de meting begint. De manchet
wordt opgeblazen. Het is normaal als demanchet erg knelt. Tijdens het opblazen
wordt
links
op
de
displayeen
drukbalkindicator weergegeven.
Opmerking: als u het opblazen op enig
moment wilt stoppen, drukt uopnieuw op de knop START .

Druk opAlle displayonderdelen in beeld

Nul-display
Start het opblazen

3.

Zodra de manchet volledig is opgeblazen,
begint het leeglopen automatisch. Het
Bezig met
hartsymbool
gaat knipperen, wat
opvoeren druk
aangeeft dat er wordt gemeten. Zodra dehartslag wordt gedetecteerd, knippert het
symbool bij elke hartslag.
Opmerking: als er geen passende druk wordt
bereikt, begint het apparaat
automatisch opnieuw met opblazen.
Raadpleeg de paragraaf “Meting met degewenste systolische druk“ op devolgende pagina als u opnieuw
opblazen wilt vermijden.
De meting wordt

4.

Zodra de meting voltooid is, worden desystolische en diastolische drukwaarden en dehartslag weergegeven.
De manchet laat de resterende lucht
ontsnappen en loopt volledig leeg.

5.

Druk op de knop START om het apparaat uit tezetten.

Opmerking: zorg voor een tussenpauze van
minstensdrie
minutenvoor
metingen bij dezelfde persoon.
Nederlands 14

START

uitgevoerd

Systolische druk
Diastolische druk
WHO-classificatie
HartslagResterende lucht wordt
automatisch uitgestoten

Metingen
Meting met de gewenste systolische druk
De UA-767S is ontworpen om de hartslag te detecteren en de manchet
automatisch tot een systolisch drukniveau op te blazen.
Gebruik de volgende methode wanneer er herhaaldelijk opnieuw wordt
opgeblazen of wanneer de resultaten niet worden weergegeven, zelfs als dedruk tot 20 mmHg of minder is gedaald.
1.

Plaats de manchet ter hoogte van het hart rond Ter hoogte van het hart
de arm (bij voorkeur de linkerarm).

2.

Druk op de knop START .

3.

Houd de knop START ingedrukt zodra denul knippert, tot een getal wordt
weergegeven dat ongeveer 30 tot 40
mmHg hoger is dan uw verwachtesystolische druk.

Laat de knop los bij degewenste systolische
druk

4.

Zodra het gewenste getal is bereikt, laatu de knop START los om de meting testarten. Ga door met het meten van uw
bloeddruk zoals beschreven op de vorige
pagina.

Houd de knop
ingedrukt

Druk op de
knop

START

START

Zie de vorige pagina voor
meting

Een nauwkeurige meting uitvoeren
Ga in een comfortabele houding zitten. Plaats uw arm op een tafel met dehandpalm omhoog en de manchet ter hoogte van uw hart.
Ontspan gedurende vijf tot tien minuten voordat u gaat meten. Als u opgewondenof depressief bent als gevolg van emotionele stress, zal de meting deze stressweerspiegelen in de vorm van een hogere (of lagere) bloeddrukwaarde dannormaal en zal de hartslag meestal sneller zijn dan normaal.
De bloeddruk van een persoon varieert constant, afhankelijk van wat u doet en
wat u hebt gegeten. Wat u drinkt kan een zeer sterk en snel effect hebben op
uw bloeddruk.
Dit apparaat baseert zijn metingen op de hartslag. Als u een erg zwakke of
onregelmatige hartslag hebt, kan het apparaat problemen hebben bij het
bepalen van uw bloeddruk.
Als het apparaat een toestand aantreft die abnormaal is, stopt het met meten
en geeft het een foutsymbool weer. Zie pagina 7 voor een beschrijving van desymbolen.
Nederlands 15

Deze bloeddrukmonitor is bedoeld voor gebruik door volwassenen. Raadpleeg
uw arts voordat u dit apparaat op een kind gebruikt. Kinderen hebben toezicht
nodig bij het gebruik van dit apparaat.
De prestatie van de automatische bloeddrukmonitor kan worden beïnvloeddoor buitensporige temperaturen, luchtvochtigheid of hoogte.

De gegevens uit het geheugen
opvragen
Opmerking: dit apparaat slaat de laatste 60 metingen op in zijn geheugen.
1. Als er niets wordt weergegeven, houdt
u de knop _START_ ingedrukt om deopgeslagen gegevens op te vragen.

Ingedrukt
houdenSTART

knop
Gemiddelde systolische druk
Gemiddelde diastolische druk

2. Laat de knop los wanneer degemiddelde waarden worden
weergegeven.

Gemiddelde hartslag

Gemiddelde vanalle gegevens

3. Het gegevensnummer en deopgeslagen gegevens worden
automatisch op volgorde weergegeven
vanaf de laatste meting.
Systolisch4. De display wordt automatischuitgeschakeld nadat alle gegevens zijn
weergegeven.

DiastolischHartslagMeest recente gegevens

Opmerking: als u tijdens het opvragen vangegevens op de knop_START_ drukt, wordt het
apparaat uitgeschakeld.

SystolischDiastolischHartslagLaatste gegevens (oudste)
Nederlands 16

Waarvoor dient de
IHB/AFib-indicator?
Zodra de bloeddrukmonitor tijdens de metingen een onregelmatige hartslag
waarneemt, wordt de IHB/AFib-indicator op de display weergegeven naast de
meetwaarden.
Opmerking: we bevelen aan dat u contact met uw huisarts opneemt als u deze
IHB/AFib-indicator regelmatig ziet.

Wat is de AFib?
Het hart trekt samen als gevolg van elektrische signalen in het hart en stuurt bloeddoor het lichaam. Atriumfibrillatie (AFib) doet zich voor wanneer het elektrische
signaal in het atrium verward raakt, wat leidt tot storingen in de hartslaginterval.
AFib kan ervoor zorgen dat bloed in het hart stagneert, wat makkelijk kan leiden
tot bloedproppen, een oorzaak voor beroertes en hartaanvallen.

％IHB/AFib%IHB/AFib wordt weergegeven als frequentie van de waargenomen IHB/AFib.
IHB/AFib kan niet alleen geluiden waarnemen, zoals fysieke bewegingen, maar
ook een onregelmatige hartslag. We raden u dan ook aan contact met uw arts
op te nemen als %IHB/AFib een hoge waarde aangeeft.
%IHB/AFib =Aantal gedetecteerde
IHB/AFib-voorvallen in geheugen

x 100 [%]

Totaal aantal

Weergave van %IHB/AFib: %IHB/AFib
wordt
weergegeven
wanneer
gemiddelde waarden worden weergegeven. (Zie “2.
De gegevens opvragen“ opvragen in de“Gebruiksmodus“)%IHB/AFib wordt niet weergegeven wanneer het
geheugennummer zes of lager is.
Weergave gemiddelde waarden

Niveau 0
Niveau 1%IHB/AFib=0 %IHB/AFib=1 - 9

Niveau 2%IHB/AFib=10 - 24

Niet
weergegeven

Nederlands 17

Niveau 3%IHB/AFib=25 - 100

Drukbalkindicator
De indicator bewaakt de drukontwikkeling tijdens metingen.
Bezig met opvoeren druk

Bezig met leeglopen

Bezig met opblazen

Bezig met meten
Opblazen is voltooid

Indicator WHO-classificatie
Elk segment van de balkindicator komt overeen
met de bloeddrukclassificatie van de WHO die
op de volgende pagina wordt beschreven.

Indicator WHO-classificatie
Ernstige hoge bloeddruk
Matige hoge bloeddruk
Milde hoge bloeddruk
Hoog-normaal
NormaalOptimaal

Voorbeeld:

Matige hoge bloeddruk

Milde hoge bloeddruk: De indicator geeft een segment weer op
basis van de huidige gegevens,corresponderend met de WHO-classificatie.

Hoog-normaal

Meer over bloeddruk
Wat is bloeddruk?
Bloeddruk is de kracht die door het bloed wordt uitgeoefend tegen de wanden
van de slagaders. Systolische druk doet zich voor wanneer het hart samentrekt.
Diastolische druk doet zich voor wanneer het hart uitzet. Bloeddruk wordt
gemeten in millimeter kwik (mmHg). Uw natuurlijke bloeddruk wordt
vertegenwoordigd door de fundamentele druk, die ‘s ochtends wordt gemeten
terwijl u nog in ruste bent en voordat u ontbijt.

Nederlands 18

Wat is hoge bloeddruk en hoe wordt deze onder controle gehouden?

Hoge bloeddruk is een abnormaal hoge slagaderlijke bloeddruk die, indien
onbehandeld, tot een groot aantal gezondheidsproblemen kan leiden,
waaronder beroertes en hartaanvallen. Hoge bloeddruk kan onder controle
worden gehouden door het veranderen van uw levensstijl, het vermijden vanstress en met medicatie onder dokterstoezicht.
U kunt het volgende doen om hoge bloeddruk te voorkomen of onder controle
te houden:
Niet roken
Regelmatig bewegen
Minder zout en vet in uw voedsel
Regelmatig een medische controle
gebruiken
ondergaan
Een gezond gewicht handhaven

Waarom thuis bloeddruk meten?

Als de bloeddruk in het ziekenhuis of bij een huisarts wordt gemeten, kan dit
spanning veroorzaken, wat tot een verhoogde waarde kan leiden, 25 tot 30 mmHg
hoger dan wanneer thuis gemeten. Thuismeting reduceert de effecten van externe
invloeden op de bloeddrukwaarden, geeft een aanvulling op de metingen van dearts en leidt tot een nauwkeuriger en meer complete bloeddrukgeschiedenis.

Bloeddrukclassificatie WHO

Bloeddrukvariaties

Diastolische bloeddruk

De Wereldgezondheidsorganisatie Referentiemateriaal: Journal of Hypertension 1999, Vol 17, nr. 2
Hoge bloeddruk type 3 (ernstig)(WHO) heeft normen opgesteld mmHg
110
voor het beoordelen van bloeddruk,
Hoge bloeddruk type 2 (gemiddeld)105
zonder onderscheid te maken naar
100
leeftijd. Deze normen worden
Hoge bloeddruk type 1 (mild)
95
weergegeven in de grafiek.
90
85
80

Hoog-normaal
Normaal

Bloeddruk

De bloeddruk van een persoon kan
Optimaal
flink variëren, zowel in de loop van
120 130 140 150 160 170 180
de dag als tijdens het seizoen. Er
Systolische bloeddruk
mmHg
kan een verschil van 30 tot 50
mmHg zijn als gevolg van diverse omstandigheden in de loop van de dag. Bij
personen met hoge bloeddruk zijn deze variaties zelfs nog groter. Normaal
gesproken stijgt de bloeddruk terwijl u werkt of aan sport doet en is deze op zijn
laagste punt wanneer u slaapt. Maak u daarom niet al te ongerust over deresultaten van één enkele meting.
Doe uw metingen elke dag op dezelfde tijd en
Normale fluctuatie binnen een dag
gebruik daarvoor de procedure die in deze(om de vijf minuten gemeten)
Systolischhandleiding wordt beschreven om uw
normalebloeddrukteachterhalen.
Regelmatige metingen geven een meer
volledige bloeddrukgeschiedenis. Zorg ervoor
dat u ook de datum en tijd opschrijft wanneeru uw bloeddruk noteert. Raadpleeg uw arts
Slaap
Diastolischom uw bloeddrukgegevens te interpreteren.
PM

Nederlands 19

AM

Tijd

Probleemoplossing
Probleem

Mogelijke oorzaak

Aanbevolen handeling
Vervang alle batterijen door
Er wordt niets op De batterijen zijn leeg.
nieuwe exemplaren.
de display
Plaats de batterijen opnieuw enweergegeven,
De contactpunten van de zorg ervoor dat de positieve
terwijl het
batterijen zitten niet op de contactpunten op de posities zitten
apparaat aan
juiste positie.
die in het batterijvak zijn
staat.
aangeduid.
Het batterijvoltage is telaag.
Het symbool
De manchet(BATTERIJ BIJNA LEEG) Vervang alle batterijen door
wordt niet
knippert. Als de batterijen nieuwe exemplaren.
opgeblazen.
helemaal leeg zijn, wordt
het symbool niet
weergegeven.
De manchet is niet correct
Breng de manchet correct aan.
aangebracht.
U hebt uw arm of lichaam Zorg ervoor dat u erg stil blijftbewogen tijdens het
zitten en geen geluid maakt tijdens
meten.
metingen.
Het apparaat
voert geen
Zorg ervoor dat u comfortabel enmeting uit.
stil zit.
De positie van de
De waarden zijn
Plaats uw arm op een tafel met demanchet is niet correct.
te hoog of te
handpalm omhoog en de manchetlaag.
ter hoogte van uw hart.
Als u een erg zwakke of
onregelmatige hartslag hebt, kan
het apparaat problemen hebben
bij het bepalen van uw bloeddruk.
De waarde wijkt af van dewaarde die bij een bezoek Zie “Waarom thuis bloeddruk
aan een ziekenhuis ofmeten?“.
huisarts is gemeten.
OverigVerwijder de batterijen. Plaats ze
op de goede manier terug en voer
de meting opnieuw uit.

Opmerking: neem contact op met de verkoper als u het probleem niet kunt
oplossen met de hierboven beschreven handelingen. Probeer
dit product niet te openen of repareren. Als u dit toch doet, wordt
uw garantie ongeldig.

Nederlands 20

OnderhoudOpen het apparaat niet. Er worden in dit apparaat delicate onderdelen en een
ingewikkelde luchteenheid gebruikt die beschadigd zouden kunnen raken. Alsu het probleem niet kunt oplossen aan de hand van deprobleemoplossingsinstructies, dient u contact op te nemen met degeautoriseerde verkoper in uw gebied of met onze afdeling klantenservice. De
klantenservice van A&D voorziet geautoriseerde verkopers van technische
informatie, reserveonderdelen en reserve-eenheden.
Het apparaat is ontworpen en gefabriceerd voor een lange levensduur. Wij
bevelen echter over het algemeen aan het apparaat elke 2 jaar te laten
inspecteren om te controleren dat dit nog goed werkt en nauwkeurig is. Neem voor
onderhoud contact op met de geautoriseerde verkoper in uw gebied of met A&D.

Technische gegevens
Type
Meetmethode
Meetbereik

Meetnauwkeurigheid
Voeding
Aantal metingen

Classificatie
Klinische test

EMD
GeheugenGebruiksomstandigheden
Omstandigheden voortransport / opslag
Afmetingen
Gewicht

UA-767S
Oscillometrische metingDruk:
0 - 299 mmHg
Systolische druk: 60 - 279 mmHg
Diastolische druk: 40 - 200 mmHg
Hartslag: 40 - 180 slagen / minuutDruk:± 3 mmHg
Hartslag: ± 5 %4 x 1,5 V-batterijen (R6P, LR6 of AA) of
netstroomadapter (TB-233C) (Niet inbegrepen)
Ongeveer 700 keer LR6 (alkalinebatterijen)
Ongeveer 200 keer R6P (mangaanbatterijen)
Met een drukwaarde van 180 mmHg bij een
kamertemperatuur van 23 °C.
ME-apparatuur met interne voeding (voeding via
batterijen) / Klasse II (voeding via adapter)
Modus continu bedrijf
Volgens ISO81060-2：2013
Tijdens de klinische validatiestudie werd K5 op 85
proefpersonen gebruikt om de diastolische bloeddruk tebepalen.
IEC 60601-1-2: 2014
Laatste 60 metingen+10 tot +40 °C / relatieve luchtvochtigheid 15 tot 85% /800 tot 1.060 hPa-20 tot +60 °C / relatieve luchtvochtigheid 10 tot 95% /700 tot 1.060 hPa
Ongeveer 140 [B] x 60 [H] x 105 [D] mm
Ongeveer Ongeveer 245 g, exclusief batterijen
Nederlands 21

Beschermingsgraad
Aangebracht onderdeel
Nuttige levensduur

Apparaat: IP20
ManchetType BF
Apparaat: 5 jaar (indien zes keer per dag gebruikt)
Manchet: 2 jaar (indien zes keer per dag gebruikt)
Netstroomadapter: 5 jaar (indien zes keer per dag
gebruikt)
Accessoire netstroomadapter
De adapter wordt gebruikt om het apparaat thuis op een voedingsbron aan tesluiten. Neem voor aankoop contact op met uw plaatselijke A&D-verkoper. De
netstroomadapter moet periodiek worden geïnspecteerd of vervangen.
TB-233C
Neem voor aankoop contact op met uw plaatselijke
A&D-verkoper.
De netstroomadapter moet periodiek wordengeïnspecteerd of vervangen.
Symbolen die op de netstroomadapter zijn gedrukt
Symbolen
Functie / betekenis
Uitsluitend voor gebruik binnenshuis
Klasse II-apparaat
Thermische zekering
Zekering
Apparaatlabel EG-richtlijn
Apparaatlabel EAC-certificatie
Polariteit connector netstroomadapter
Accessoires worden afzonderlijk verkocht
Manchet Catalogusnummer
Manchetgrootte
Armomvang
CUF-F-LA
Grote manchet volwassenen
31 cm tot 45 cm
CUF-I
Manchet voor breed spectrum 22 cm tot 42 cm
CUF-F-A
Manchet volwassenen
22 cm tot 32 cm
Armomvang: de omtrek van de arm ter hoogte van de biceps.
Netstroomadapter

Catalogusnummer
TB-233C

Stekker (type wandcontact)
Type C

Opmerking: specificaties kunnen zonder voorafgaande kennisgeving worden
gewijzigd.
De IP-classificatie is de mate van bescherming die door behuizingen
wordt geboden conform IEC 60529. Dit apparaat is beschermd tegenvaste vreemde voorwerpen met een diameter van 12 mm of groter,zoals vingers. Dit apparaat is niet beschermd tegen water.

Nederlands 22

IndholdKære kunder ......................................................................... 2
Indledende bemærkninger .................................................... 2
Sikkerhedsforholdsregler ...................................................... 2
Oversigt over dele................................................................. 5
Symboler .............................................................................. 6
Driftstilstand ........................................................................ 8Sådan bruges måleren .......................................................... 9Isætning/udskiftning af batterier ..................................... 9
Tilslutning af luftslangen ............................................... 10
Tilslutning af AC-adapteren ........................................... 10
Valg af den korrekt manchetstørrelse ............................. 11Påsætning af arm-manchetten ....................................... 11Sådan foretager du nøjagtige målinger........................... 13Måling ........................................................................... 13
Efter måling ................................................................... 13Målinger ............................................................................. 14
Normal måling ............................................................... 14Måling med det ønskede systoliske tryk ......................... 15Bemærkninger om nøjagtig måling ................................ 15
Hentning af hukommelsesdata ............................................ 16
Hvad er indikatoren for IHB/AFib? ....................................... 17
Hvad er AFib? ...................................................................... 17％IHB/AFib .......................................................................... 17
Tryksøjleindikator ............................................................... 18
WHO-klassifikationsindikator ............................................. 18
Om blodtryk ....................................................................... 18
Hvad er blodtryk? ........................................................... 18
Hvad er hypertension, og hvordan kontrolleres det? ....... 19
Hvorfor måle blodtryk derhjemme? ................................ 19
WHO’s blodtryksklassifikation ........................................ 19
Variationer i blodtryk ..................................................... 19
Fejlfinding .......................................................................... 20
Vedligeholdelse .................................................................. 21
Tekniske data ..................................................................... 21

Dansk 1Kære kunder
Tillykke med dit køb af denne A&D blodtryksmåler. Dette apparat er udviklet
med fokus på brugervenlighed og nøjagtighed, og det vil lette dine daglige
aktiviteter omkring blodtryk.
Vi anbefaler, at du gennemlæser denne manual omhyggeligt, før dubruger apparatet for første gang.

Indledende bemærkninger
Dette apparat overholder det europæiske direktiv 93/42 EØF for medicinske
produkter. Dette markeres med-overensstemmelsesmærkatet.
(0123: referencenummeret for det relevante bemyndigede organ)
Apparatet er beregnet til brug på voksne, ikke på nyfødte eller for tidligt
fødte børn.
Anvendelsessteder. Apparatet er beregnet til din egen private brug i hjemmet.
Apparatet er udviklet til at måle blodtryk og pulsfrekvens på mennesker med
henblik på diagnose.

Sikkerhedsforholdsregler
Apparatet er fremstillet med brug af præcisionskomponenter. Store udsving itemperatur og luftfugtighed samt direkte sollys, stød og støv skal undgås.
Rengør apparatet og manchetten med en tør, blød klud eller en klud, der erfugtet med vand og et neutralt rengøringsmiddel. Brug aldrig sprit, benzen,
fortynder eller andre stærke kemikalier til rengøring af apparatet eller
manchetten.
Undgå at folde manchetten stramt eller opbevare slangen i længere tid, mensden er vredet eller snoet, da det kan forringe levetiden for apparatets dele.
Vær omhyggelig med at forhindre, at spædbørn eller små børn kan blive
kvalt af slangen og kablet.
Luftslangen må ikke vrides under måling. Det kan medføre personskade på
grund af længerevarende tryk fra manchetten.
Apparatet og manchetten er ikke vandtætte. Undgå, at der kommer regn,
sved eller vand på apparatet og manchetten.
Målingerne kan forstyrres, hvis apparatet anvendes tæt på TV-apparater,
mikrobølgeovne, mobiltelefoner, røntgenapparater eller andre apparater
med stærke elektriske felter.
Trådløse kommunikationsenheder, f.eks. netværksenheder, mobiltelefoner,trådløse telefoner og deres basestationer samt walkie-talkies, kan påvirke
blodtryksmåleren. Hold derfor en afstand på mindst 30 cm fra sådanne
apparater.
Dansk 2

Ved genbrug af apparatet skal du kontrollere, at apparatet er rent.
Brugt udstyr, dele og batterier må ikke behandles som almindeligt
husholdningsaffald, men skal bortskaffes i overensstemmelse medgældende lokale regler og bestemmelser.
Når AC-adapteren anvendes, skal du sørge for, at den let kan tages ud af
stikkontakten, når det er nødvendigt.
Der må ikke foretages ændringer af apparatet. Det kan forårsage i ulykker
eller skader på apparatet.
For at måle blodtrykket skal armen klemmes kraftigt nok af manchetten til
midlertidigt at stoppe blodgennemstrømningen gennem arterien. Det kanforårsage smerter, følelsesløshed eller et midlertidig rødt mærke på armen.
Denne tilstand kan især opstå, hvis målingen gentages et antal gange.
Eventuel smerte, følelsesløshed eller røde mærker vil forsvinde med tiden.
Overdrevent hyppig måling af blodtrykket kan forårsage skader, fordi der gribes
ind i blodgennemstrømningen. Kontroller, at brugen af apparatet ikke forringer
blodcirkulationen i længere tid, hvis apparatet bruges gentagne gange.
Hvis du har fået udført en mastektomi (fjernelse af brystet), skal du rådføre
dig med en læge, før du bruger apparatet.
Lad ikke børn bruge apparatet selv, og undgå at bruge apparatet på et sted,hvor små børn har adgang. Det kan forårsage ulykker eller skader.
Apparatet indeholder små dele, som kan udgøre kvælningsfare, hvis etspædbarn sluger en del ved et uheld.
Tag AC-adapteren ud af stikkontakten, når den ikke er i brug under målingen.
Brug af tilbehør, som ikke er beskrevet i denne manual, kan forringe
sikkerheden.
Hvis batteriet skulle blive kortsluttet, kan det blive varmt og eventueltforårsage forbrændinger.
Lad apparatets temperatur tilpasse sig til omgivelserne før brug (cirka en time).
Der har ikke været udført klinisk test på nyfødte børn og gravide kvinder. Må
ikke bruges på nyfødte spædbørn eller gravide kvinder.
Undgå at røre ved batterierne, DC-stikket og patienten på samme tid. Det
kan medføre elektrisk stød.
Undgå at pumpe op, uden at manchetten er sat omkring overarmen.

Kontraindikationer

Følgende sikkerhedsforholdsregler skal overholdes for korrekt brug af apparatet.
Manchetten må ikke sættes på en arm, hvor der er påsat andet medicinsk
elektrisk udstyr. Udstyret vil måske ikke fungere korrekt.
Personer med betydeligt nedsat kredsløb i armen skal rådføre sig med enlæge, før apparatet bruges, for at undgå helbredsproblemer.
Lad være med selv at stille diagnoser ud fra måleresultater og selv starte en
behandling. Kontakt altid din læge med henblik på vurdering af resultaterneog eventuel behandling.
Manchetten må ikke sættes på en arm, der har et ikke-helet sår.
Dansk 3

Manchetten må ikke sættes på en arm, der modtager intravenøst drop eller
blodtransfusion. Det kan medføre personskade eller ulykker.
Apparatet må ikke bruges, hvor der er brændbare gasser, f.eks.
anæstesigasser. Det kan medføre eksplosion.
Apparatet må ikke anvendes i omgivelser med høj iltkoncentration, foreksempel et iltkammer eller et ilttelt. Det kan medføre brand eller eksplosion.

Dansk 4

Oversigt over deleSkærm

LuftstikAC-adapterstik
DC-stik
Knappen START

LuftfatningOmråde for korrekt
pasform

Luftslange
Positionsmærke for arterie

Indexmærke
Arm-manchet
Batterirum
Batteridæksel

1,5 V-batterier(R6P, LR6 eller AA)Skærm

HUKOMMELSE

Gennemsnit

Symbol for IHB/AFib
Systolisk
tryk%IHB/AFib
Symbol for manchet, der ikke passer

Diastolisk
tryk

Symbol for bevægelsesfejl
Hjertemærke
WHO-klassifikationsindikator
og tryksøjleindikator

Pulsfrekvens

Batteriindikator
Dansk 5

Symboler
Symboler, der er trykt på apparatets kabinet
Symboler
SYS
DIA
PUL

Funktion/betydningSæt apparatet på standby, og tænd det igen.
Systolisk blodtryk i mmHg
Diastolisk blodtryk i mmHg
Pulsslag pr. minut
Installationsvejledning for batteriJævnstrøm
Type BF: Apparat, manchet og slanger er designet til at give særlig
beskyttelse mod elektrisk stød.
Mærke for EU-direktiv på medicinsk udstyr
EU-repræsentant
ProducentFremstillingsdato

IPInternationalt beskyttelsessymbol
WEEE-mærkat

SN

Serienummer
Se brugermanual/hæfte
Polaritet for DC-stik
Opbevares tørt

Symboler, der vises på skærmen
Symboler

Funktion/betydningAnbefalet handling
Vises, mens måling er i gang. Blinker,Måling er i gang. Undgå
når der registreres puls.
bevægelser.
Symbol for IHB/AFib
Vises, når der registreres uregelmæssig
hjerterytme.
Det kan tændes, når der registreresvibrationer i form af skælven ellerrystelse.
Udlæsningen kan haveen forkert værdi.
Vises, når der registreres bevægelse af Prøv at foretage
kroppen eller armen.
målingen igen. Undgå
bevægelser undermålingen.
Dansk 6

Symboler
Symboler

Funktion/betydning
Vises under måling, når manchetten
er monteret løst

Anbefalet handlingUdlæsningen kan have en
forkert værdi. Sætmanchetten korrekt på, og
prøv at foretage målingen
igen.

Registreret IHB/AFib-frekvens i
hukommelsen%IHB/AFib =Antal registrerede
IHB/AFibs i hukommelsen x100 [%]Samlet antal

Tidligere målinger lagret i
hukommelsen.
Gennemsnitsdata

FULDT BATTERINIVEAU
Indikatoren for batteristrøm undermåling.
LAVT BATTERINIVEAU
Udskift alle batterier med
Batterikapacitet er lav, når den blinker. nye, når mærket blinker.
Prøv at foretage målingen
igen.
Ustabilt blodtryk på grund afbevægelse under måling.
Man skal undgå at bevæge
sig under målingen.
De systoliske og diastoliske værdier
ligger inden for 10 mmHg fra
hinanden.
Trykværdien steg ikke underSæt manchetten korrekt
oppumpning.
på, og prøv at foretage
målingen igen.
Manchetten er ikke sat korrekt på.
PULSFEJL
Pulsen blev ikke registreret korrekt.

Intern fejl i blodtryksmåleren

Dansk 7

Tag batterierne ud, tryk på
knappen START , og sæt
derefter batterierne i igen.
Kontakt forhandleren, hvis
fejlen stadig vises.

Driftstilstand1. Normal måling
Tryk på knappen START . Blodtrykket måles, og dataene lagres i
hukommelsen. Apparatet kan gemme de sidste 60 målinger i hukommelsen.

2. Hentning af data
Hvis der ikke vises noget, skal du trykke på
knappen START og holde den inde.

I standbyskal du trykke påog holde knappen
START

Slip knappen, når gennemsnitsdataene vises.

Datanummeret og de lagrede data vises
automatisk i rækkefølge fra den seneste måling.

Du kan finde flere oplysninger om at hente data
under “Hentning af hukommelsesdata“.

3. Sletning af alle data, der er lagret i hukommelsen
Tryk på knappen
START , mens du slukker apparatet, indtil “Vælg ““ for at slette dataene.
blinker.
Dataene er slettet, når mærket
Apparatet slukkes automatisk.

“ vises.

4. Måling med det ønskede systoliske tryk
Se side 15 for måling med det ønskede systoliske tryk.

Dansk 8STARTSådan bruges målerenIsætning/udskiftning af batterier
1. Fjern batteridækslet.
2. Fjern de brugte batterier, og sæt nye
batterier i batterirummet som vist.
Sørg for, at polerne (+ og -) venderkorrekt.
Brug kun batterier af typen R6P,
LR6 eller AA.
3. Sæt batteridækslet på.

Trin 1
Trin 3

Trin 1
Trin 2FORSIGTIGSæt batterierne i batterirummet som vist. Hvis de sættes forkert i, fungerer
apparatet ikke.
Når(mærke for LAVT BATTERINIVEAU) blinker på skærmen, skal allebatterierne udskiftes med nye. Brug ikke gamle og nye batterier sammen. Det
kan forkorte batteriets levetid eller få apparatet til at fungere forkert.
Vent mindst to sekunder efter at have slukket apparatet, før du udskifter
batterierne.
(mærke for LAVT BATTERINIVEAU) stadig vises, efter at
Hvis
batterierne er udskiftet, skal du foretage en blodtryksmåling. Derved kan
apparatet registrere de nye batterier.
(mærke for LAVT BATTERINIVEAU) vises ikke, når batterierne er tomme.
Batteriets levetid afhænger af den omgivende temperatur og kan være kortere
ved lave temperaturer. Generelt holder fire nye LR6-batterier cirka et år ved todaglige målinger.
Brug kun de specificerede batterier. Batterierne, der følger med apparatet, er
til test af målerens funktioner og kan have en begrænset levetid.
Fjern batterierne, hvis apparatet ikke skal bruges i længere tid.
Batterierne kan blive utætte og medføre funktionsfejl.

Dansk 9Sådan bruges måleren
Tilslutning af luftslangen

Luftfatning

Sæt luftstikket godt fast i
luftfatningen.

LuftstikTilslutning af AC-adapterenSæt AC-adapterstikket i DC-stikket.
Forbind derefter AC-adapteren til en
stikkontakt.
Brug den specificerede AC-adapter.
(Se side 22.)

DC-stikAC-adapterstik

Når du tager AC-adapteren ud af stikkontakten, skal du gribe om
AC-adapterens hus og trække det ud af stikkontakten.
Når du tager AC-adapterstikket ud af blodtryksmåleren, skal du gribe om
AC-adapterstikket og trække det ud af måleren.

Dansk 10Sådan bruges måleren
Valg af den korrekt manchetstørrelse
En korrekt manchetstørrelse er vigtig for en nøjagtig måling. Hvis manchetten
ikke har den rigtige størrelse, kan målingen vise en forkert blodtryksværdi.
Armstørrelsen er trykt på hver manchet.
Index
og område for korrekt pasform på manchetten fortæller dig, om dubruger den rigtige manchet. (Se “symboler, der er trykt på manchetten” pånæste side.)
Hvis index
peger uden for området, skal du kontakte din lokale forhandler
for at købe en anden manchet.
Arm-manchetten er en forbrugsvare. Køb en ny, hvis det bliver slidt.
Armstørrelse
Anbefalet manchetstørrelse
Katalognummer
31 cm til 45 cm
Stor manchet til voksne
CUF-F-LA
22 cm til 42 cm
Universalmanchet
CUF-I
22 cm til 32 cm
Manchet til voksne
CUF-F-A
Armstørrelse: omkredsen ved biceps.
Note:
UA-767S er ikke beregnet til brug af en lille manchet.

Påsætning af arm-manchetten
1. Festmanchetten
rundtom
overarmen, ca. 1-2 cm overindersiden af albuen som vist.
Sæt manchetten direkte mod huden,da tøj kan gøre pulsen svagere, såder opstår fejl i målingen.

Rul ikke skjorteærmet for stramt op.

2. Stramning omkring overarmen på
grund af et stramt oprullet
skjorteærme kan forhindre nøjagtigemålinger.

Positionsmærke for arterie

Manchet1-2 cmLuftslange

Klem stoffet
fast til.

3. Kontroller, at index
peger inden
for området for korrekt pasform.
Index

Dansk 11og område for korrekt pasformSådan bruges måleren
Symboler, der er trykt på manchetten
Symboler Funktion/betydning
Positionsmærke for arterie▲
REF

Anbefalet handlingpåSætmærketindersiden af armen ved
arterien i overarmen, ellerså den flugter med
ringfingeren.

Index
KatalognummerOmråde for korrekt pasform formanchet til voksne.
Dette er trykt på manchetten til
voksne.
Brug stor manchet til
Over området, trykt på manchet til voksneistedetforvoksne og universalmanchet.
manchet til voksne eller
universalmanchet.
Område for korrekt pasform foruniversalmanchet. Dette er trykt påuniversalmanchetten.
Område for korrekt pasform for stor
manchet til voksne. Dette er trykt på
den store manchet til voksne.
Under området, trykt på manchet til
voksne og universalmanchet.
Brug manchet til voksne i
Under området, på stor manchet til
stedet for stor manchet til
voksne.
voksne.
Partinummer

Stor manchet til voksneAOmråde for korrekt pasform

L

Universalmanchet

S

W

L

A

L

Manchet til voksneS

Dansk 12Sådan bruges målerenSådan foretager du nøjagtige målinger
Gør følgende for at få den mest præcise blodtryksmåling:Sæt dig i en stol, så du sidder behageligt. Hvil armen på bordet. Sid ikke med
krydsede ben. Lad fødderne hvile fladt på gulvet, og ret ryggen.
Slap af i ca. fem til ti minutter før målingen.
Anbring midten af manchetten i samme højde som dit hjerte.
Sid stille, og tal ikke under målingen.
Undlad at måle lige efter fysisk anstrengelse eller badning. Hvil dig i tyve eller
tredive minutter før målingen.
Prøv at måle dit blodtryk på samme tid hver dag.

MålingDet er normalt, at manchetten føles meget stram under målingen. (ingen grund
til at være urolig.)

Efter måling
Efter målingen skal du trykke på knappen START
Fjern manchetten, og skriv dine data ned.

for at slukke for apparatet.

Bemærk: Apparatet har en funktion til automatisk slukning, som afbryder
apparatet cirka et minut efter målingen.
Der skal gå mindst tre minutter mellem målinger på den samme person.

Dansk 13Målinger
Se “Bemærkninger om nøjagtig måling“ på næste side, før du begynder påmålingen.

Normal måling
1.

Sæt manchetten på armen (helst venstre arm).
På højde med hjertet
Sid stille under målingen.

2.

Tryk på knappen START .
Alle skærmsegmenterne vises.
Tryk på
Nul (0) vises og blinker kortvarigt.
Skærmen skifter som vist i figuren til højre,når målingen starter. Manchetten begynder
Alle skærmsegmenterne vises
at blive pumpet op. Det er normalt, atmanchetten føles meget stram. En
tryksøjleindikator vises i venstre side afskærmen under oppumpningen.
Bemærk: Hvis du vil stoppe oppumpningen,Nul-visningkan du til enhver tid trykke på Starter oppumpningknappen START igen.

3.

Når oppumpningen er færdig, startertømningen af luft automatisk, og(hjertemærke) blinker som angivelse af, atmålingen er i gang. Når pulsen registreres,
blinker mærket med hvert pulsslag.
Bemærk: Hvis der ikke opnås et passendetryk, starter apparatet automatisk
på oppumpning igen.
Hvis du vil undgå genoppumpning,kan du se “Måling med det ønskede
systoliske tryk“ på næste side.

4.

Når målingen er fuldført, vises de systoliske og
diastoliske trykmålinger samt pulsfrekvensen.
Manchetten tømmes helt for resterende luft.

5.

Tryk på knappen
apparatet.

START

START

Øger trykket

Måling i gang

for at slukke forBemærk: Der skal gå mindst tre minutter
mellem målinger på den samme
person.

Dansk 14

Systolisk tryk
Diastolisk tryk
WHO-klassifikation
Pulsfrekvens
Lukker automatisk
resterende luft udMålingerMåling med det ønskede systoliske tryk
UA-767S er konstrueret, så den registrerer pulsen og automatisk pumper
manchetten op til systolisk trykniveau.
Brug denne metode, hvis der ofte forekommer genoppumpning, eller hvis
der ikke vises resultater, selvom trykket falder til 20 mmHg eller derunder.
1.

Sæt manchetten på armen i højde medPå højde med hjertet
hjertet (helst venstre arm).

2.

Tryk på knappen

3.

Mens nullet blinker, skal du trykke på
knappen START og holde den inde,indtil der vises et tal, der er cirka 30 til
40 mmHg højere end det forventede
systoliske tryk.

4.

Når det ønskede tal er nået, skal duslippe knappen START for at startemålingen. Fortsæt for at måle dit
blodtryk som beskrevet på forrige side.

Tryk på START
knappenSTART .

Slip knappen ved
det ønskede
systoliske tryk

Tryk på knappen,og hold den inde

START

Se den forrige side om måling

Bemærkninger om nøjagtig måling
Sæt dig i en behagelig stilling. Læg armen på et bord med håndfladen opadog manchetten i samme højde som dit hjerte.
Slap af i ca. fem til ti minutter, før du tager en måling. Hvis du føler diganspændt eller følelsesmæssigt belastet, vil målingen afspejle dette stresssom en blodtryksvisning, der er højere (eller lavere) end den normale
blodtryksmåling, og pulsen vil som regel være hurtigere end normalt.
Dit blodtryk varierer hele tiden, afhængigt af hvad du foretager dig, og hvad
du har spist. Det, du drikker, kan have stor og hurtig virkning på dit blodtryk.
Dette apparat baserer sine målinger på pulsen. Hvis du har en meget svag
puls eller uregelmæssig hjerterytme, kan apparatet have svært ved at finde
dit blodtryk.
Hvis apparatet skulle registrere en unormal tilstand, stopper det målingen,og der vises et fejlsymbol. Se side 7 for at få en beskrivelse af symbolerne.

Dansk 15

Denne blodtryksmåler er kun beregnet til brug af voksne. Rådfør dig med dinlæge, før du bruger apparatet på et barn. Et barn må ikke bruge apparatet
uden opsyn.
Den automatiske blodtryksmålers ydeevne kan påvirkes af for høj
temperatur, luftfugtighed eller højde over vandets overflade.

Hentning af hukommelsesdataBemærk: Dette apparat lagrer de sidste 60 målinger i hukommelsen.
1. Hvis der ikke vises noget, skal du
trykke på knappen _START_ og
holde den inde for at hente delagrede data.

Tryk på og START
hold
Gennemsnitlig systolisk
Gennemsnitlig diastolisk
Gennemsnitlig puls

Gennemsnit foralle dataene

2. Slip knappen, nårgennemsnitsdataene vises.

3. Datanummeret og de lagrede datavises automatisk i rækkefølge fra denseneste måling.
Systolisk

4. Skærmen slukkes automatisk, efter atalle data er vist.

Diastolisk
Puls
Seneste dataBemærk: Hvis du trykker på knappen_START_ , mens du henter
dataene, slukkes apparatet.

Systolisk
Diastolisk
Puls
Sidste data (ældste)

Dansk 16

Hvad er indikatoren for IHB/AFib?
Hvis måleren registrerer en uregelmæssig rytme under målingerne, vises
indikatoren for IHB/AFib på skærmen med måleværdierne.
Bemærk: Vi anbefaler, at du kontakter din læge, hvis du ofte ser denne.
indikator for IHB/AFib

Hvad er AFib?
Hjertet trækker sig sammen på grund af elektriske signaler i hjertet og
sender blod gennem kroppen. Atrieflimren (AFib) opstår, når det elektriske
signal i hjertets forkammer bliver forstyrret og fører til ændringer i
pulsintervallet. AFib kan få blod til at blive i hjertet, så der let kan dannes
blodpropper, som kan forårsage slagtilfælde og hjerteanfald.

％IHB/AFib%IHB/AFib vises som hyppigheden af den registrerede IHB/AFib.
%IHB/AFib kan ikke kun registrere støj som f. eks. fysiske bevægelser, menogså en uregelmæssig hjerterytme. Derfor anbefaler vi, at du kontakter dinlæge, hvis %IHB/AFib-niveauet er højt.
Antal registrerede
IHB/AFibs i hukommelsen
x 100 [%]
%IHB/AFib =Samlet antal
Visning af %IHB/AFib: %IHB/AFib vises under visning af gennemsnitsværdier.
(Se “2. Hentning af data“ i “Driftstilstand“)%IHB/AFib vises ikke, når hukommelsesnummeret er
seks eller derunder.
Visning af gennemsnitsværdi

Niveau 0
Niveau 1
Niveau 2
Niveau 3%IHB/AFib=0 %IHB/AFib=1 - 9 %IHB/AFib=10 - 24 %IHB/AFib=25 - 100
Vises ikke

Dansk 17

Tryksøjleindikator
Indikatoren følger status for trykket under måling.
Øger trykket

Lukker luft ud

Oppumpning i gangMåling i gang
Oppumpning fuldført

WHO-klassifikationsindikator
Hvert segment i søjleindikatoren svarer til
WHO’s blodtryksklassifikation, som er
beskrevet på næste side.
WHO-klassifikationsindikatorSvær hypertension
Moderat hypertension
Mild hypertension
Høj normal
NormalOptimalEksempel:

Moderat hypertension

Mild hypertension: Indikatoren viser et segment baseret på deaktuelle data, svarende til
WHO-klassifikationen.
Høj normal

Om blodtryk
Hvad er blodtryk?
Blodtryk er den kraft, som blodet udøver mod væggene i pulsårerne. Det
systoliske tryk opstår, når hjertet trækker sig sammen. Det diastoliske trykopstår, når hjertet udvider sig. Blodtrykket måles i millimeter kviksølv(mmHg). En persons naturlige blodtryk er det grundlæggende tryk, dermåles som det første om morgenen, mens man stadig er i hvile og ikke har
spist.

Dansk 18

Hvad er hypertension, og hvordan kontrolleres det?
Hypertension, et unormalt højt arterielt blodtryk, kan forårsage mange
helbredsproblemer, herunder slagtilfælde og hjerteanfald, hvis det ignoreres.
Hypertension kan kontrolleres ved at ændre livsstil, så man undgår stress,
og med medicin under opsyn af en læge.
Sådan kan du forebygge hypertension eller holde den under kontrol:
Ryg ikke
Dyrk regelmæssig motion
Spis mindre salt og fedt
Få dit helbred tjekket regelmæssigt
Hold den rigtige vægtHvorfor måle blodtryk derhjemme?
Blodtryksmåling i en klinik eller hos en læge kan skabe ængstelse og giveen højere måling, 25 til 30 mmHg højere end målt derhjemme. Måling ihjemmet reducerer udefra kommende påvirkninger af blodtryksmålingen,den supplerer lægens målinger og giver en mere nøjagtig og komplet
historik over blodtrykket.

WHO’s blodtryksklassifikation

Variationer i blodtryk

Referencemateriale: Journal of Hypertension 1999, Vol 17 No.2
mmHg
Grad 3 hypertension (svær)110
Grad 2 hypertension (moderat)
105
Diastolisk blodtryk

Verdenssundhedsorganisationen (WHO) har udviklet
standarder for vurdering af
forhøjet blodtryk, uafhængigt
af alderen, se diagrammet.

100
95
90
85
80

Grad 1 hypertension (mild)
Høj-normal

Blodtryk

En persons blodtryk varierer
NormalOptimalmeget i løbet af dagen og fra
120 130 140 150 160 170 180årstid til årstid. Det kan variere
Systolisk blodtryk
i mmHgmed 30 til 50 mmHg på grund
af forskellige omstændigheder i løbet af dagen. Hos personer med
hypertension er variationerne endnu mere udtalte. Normalt stiger
blodtrykket, når man arbejder eller leger, og falder til det laveste niveau, når
man sover. Bliv derfor ikke unødvendigt bekymret over resultaterne af enenkelt måling.
Typiske udsving i løbet af dagen(Målt hvert femte minut)
Tag målingerne på samme tid hver dag, og
Systolisk
følg proceduren i denne manual for at
kende dit normale blodtryk. Regelmæssigemålinger giver en mere omfattende historik
over blodtrykket. Husk at skrive dato og
klokkeslæt, når du registrerer dit blodtryk.
Søvn
DiastoliskRådfør dig med din læge for at fortolke
dine blodtryksdata.
PM
AM
KlokkenDansk 19

Fejlfinding
Problem
Der vises intet
på skærmen,
selvomapparatet ertændt.

Mulig årsag
Batterierne er tømt.

Anbefalet handling
Udskift alle batterierne med nye.

Batteripolerne er ikke
placeret korrekt.

Sæt batterierne i, sådan at denegative og positive poler svarer
til det, der vises i batterirummet.

Batterispændingen er forlav.
(mærke for LAVTManchetten
BATTERINIVEAU)
pumpes ikke op.
blinker. Hvis batterierne
er tømt helt, visesmærket ikke.
Manchetten er ikke sat
rigtigt på.
Du har bevæget armeneller kroppen under
Apparatet måler målingen.
ikke.
Målingerne er
Manchetten er ikke
for høje eller for placeret korrekt.
lave.

Andet

Værdien er forskellig fra
det, der blev målt på en
klinik eller hos en læge.

Udskift alle batterierne med nye.

Sæt manchetten korrekt på.
Sørg for at sidde meget stille, og
tal ikke under målingen.
Sid behageligt og stille.
Læg armen på et bord medhåndfladen opad og manchetten isamme højde som dit hjerte.
Hvis du har en meget svag puls
eller uregelmæssig hjerterytme,
kan apparatet have svært ved at
finde dit blodtryk.
Se “Hvorfor måle blodtryk
derhjemme?“.

Tag batterierne ud. Sæt dem
korrekt i, og prøv at foretage
målingen igen.
Bemærk: Hvis de ovenstående handlinger ikke løser problemet, bedes dukontakte forhandleren. Forsøg ikke at åbne eller reparere dette produkt,
da ethvert forsøg på at gøre det vil gøre garantien ugyldig.

Dansk 20

VedligeholdelseUndgå at åbne apparatet. Det bruger følsomme, elektriske komponenter og
en kompliceret luftenhed, der kan blive beskadiget. Hvis du ikke kan løse
problemet ved hjælp af vejledning til fejlfinding, bedes du kontakte en
autoriseret forhandler i dit område eller vores kundeserviceafdeling. A&D’s
kundeservice leverer tekniske oplysninger, reservedele og apparater til
autoriserede forhandlere.
Apparatet er designet og fremstillet til at have en lang levetid. Generelt
anbefales det dog at få apparatet efterset hvert 2. år for at sikre korrekt
funktion og nøjagtighed. Kontakt en autoriseret forhandler i dit område eller
A&D for vedligeholdelse.

Tekniske dataTypeMålemetodeMåleområde

UA-767S
Oscillometrisk måling
Tryk:
0 - 299 mmHg
Systolisk tryk: 60 - 279 mmHg
Diastolisk tryk: 40 - 200 mmHg
Puls:
40 - 180 slag/minutMålenøjagtighed
Tryk:±3 mmHgPuls:±5 %Strømforsyning
4 x 1,5 V-batterier (R6P, LR6 eller AA) eller
AC-adapter (TB-233C) (Medfølger ikke)
Antal målinger
Ca. 700 gange
LR6 (alkaliske batterier)Ca. 200 gange
R6P (manganbatterier)
Ved en trykværdi på 180 mmHg og en
rumtemperatur på 23 °C.
Klassifikation
Internt strømforsynet udstyr (forsynes fra batterier) /Klasse II (forsynes fra adapter)
Kontinuerlig driftstilstand
Klinisk test
I henhold til ISO81060-2:2013
I det kliniske valideringsstudie blev K5 anvendt til 85
deltagere for at fastslå det diastoliske blodtryk.
EMI
IEC 60601-1-2: 2014
Hukommelse
Sidste 60 målinger
Driftsforhold+10 til +40 °C / 15 til 85 % RH / 800 til 1.060 hPa
Transport / opbevaringsforhold -20 til +60 °C / 10 til 95 % RH / 700 til 1.060 hPa
Dimensioner
Ca. 140 [B] x 60 [H] x 105 [D] mm
VægtCa. 245 g uden batterier
Indtrængningsbeskyttelse
Apparat: IP20
Dansk 21Påsat delForventet levetid

ManchetType BF
Apparat:5 år (ved brug seks gange om dagen)Manchet: 2 år (ved brug seks gange om dagen)AC-adapter: 5 år (ved brug seks gange om dagen)Tilbehør AC-adapter
Adapteren bruges til at forbinde apparatet til en strømkilde i hjemmet. Kontakt
din lokale A&D-forhandler ved køb. AC-adapteren skal efterses eller udskiftes
med jævne mellemrum.
TB-233C
Kontakt din lokale A&D-forhandler vedrørende køb.
AC-adapteren skal efterses eller udskiftes medjævne mellemrum.
Symboler, der er trykt på AC-adapteren
Symboler
Funktion/betydning
Kun til indendørs brug
Klasse II-apparat
Termosikring
SikringMærke for EU-direktiv på apparatet
EAC-certificeringsmærke på apparatet
Polaritet for AC-adapterstik
Tilbehør, som sælges separat
Manchet Katalognummer
Manchetstørrelse
Armstørrelse
CUF-F-LA
Stor manchet til voksne
31 cm til 45 cm
CUF-I
Universalmanchet
22 cm til 42 cm
CUF-F-A
Manchet til voksne
22 cm til 32 cm
Armstørrelse: omkredsen ved biceps.
AC-adapter

Katalognummer
TB-233C

Stik (stikkontakttype)Type CBemærk: Specifikationerne kan ændres uden forudgående varsel.
IP-klassificering er beskyttelsesgraden, som ydes af kabinetter ioverensstemmelse med IEC 60529. Dette apparat er beskyttet mod
faste fremmedlegemer på 12 mm i diameter og derover, f.eks. fingre.
Dette apparat er ikke beskyttet mod vand.

Dansk 22

Innehållsförteckning
Bästa kund ........................................................................................... 2
Inledande anmärkningar ...................................................................... 2
Försiktighetsåtgärder ........................................................................... 2Ingående delar ..................................................................................... 5
Symboler ............................................................................................. 6
Driftsläge ............................................................................................. 8Använda mätaren ................................................................................. 9
Installera/byta batterier .................................................................. 9
Ansluta luftslangen ...................................................................... 10
Ansluta nätadaptern ..................................................................... 10
Välja rätt manschettstorlek ........................................................... 11
Sätta på armmanschetten ............................................................. 11Så här görs korrekta mätningar .................................................... 13
Mätning ....................................................................................... 13
Efter mätningen............................................................................ 13
Mätningar .......................................................................................... 14
Normal mätning ........................................................................... 14
Mäta med önskat systoliskt tryck .................................................. 15
Anmärkningar för korrekt mätning ............................................... 15
Hämta minnesdata ............................................................................. 16
Vad är IHB/AFib-indikatorn? ............................................................... 17
Vad är AFib (förmaksflimmer)? ........................................................... 17%IHB/AFib (arytmi/förmaksflimmer) ................................................... 17
Tryckindikator ................................................................................... 18
Indikator för WHO-klassificering ........................................................ 18
Om blodtryck ..................................................................................... 18
Vad är blodtryck? ......................................................................... 18
Var är hypertoni och hur kontrolleras det? .................................... 19
Varför mäta blodtrycket i hemmet? ............................................... 19
WHO:s blodtrycksklassificering ..................................................... 19
Variationer i blodtrycket ............................................................... 19
Felsökning ......................................................................................... 20
Underhåll ........................................................................................... 21
Tekniska data .................................................................................... 21

Svenska 1

Bästa kund
Gratulerar till ditt inköp av en toppmodern blodtrycksmätare från A&D, en
av de mest avancerade mätarna på marknaden idag. Enheten är
utformad för att vara enkel att använda och samtidigt vara noggrann och
underlätta din dagliga blodtrycksmätningsrutin.
Vi rekommenderar att du läser igenom bruksanvisningen noggrant
innan du använder enheten för första gången.

Inledande anmärkningar
Enheten överensstämmer med Rådets direktiv 93/42/EEG om-symbolen för
medicintekniska produkter. Det indikeras genomöverensstämmelse. (0123: Referensnumret avser det aktuella anmälda
organet)Enheten är utformad för användning på vuxna, inte på nyfödda ellersmåbarn.
Användningsmiljö. Enheten är avsedd för personligt bruk för vård i hemmet.
Enheten är avsedd för att mäta blodtrycket och pulsfrekvensen hos
människor i diagnostiskt syfte.

Försiktighetsåtgärder
Precisionskomponenter har använts vid tillverkningen av enheten. Undvik
extrema temperaturer, luftfuktighet, direkt solljus, stötar och damm.
Rengör enheten och manschetten med en torr, mjuk trasa eller en trasa
fuktad med vatten och ett neutralt rengöringsmedel. Använd aldrig
alkohol, bensen, thinner eller andra aggressiva kemikalier för att rengöra
enheten eller manschetten.
Undvik att vika ihop manschetten för hårt och förvara inte slangen hårtlindad under längre tidsperioder eftersom det kan förkorta
komponenternas livslängd.
Vidta åtgärder så att det inte finns risk för att spädbarn och små barn
stryps av slangen och kabeln.
Vrid inte luftslangen under mätningen. Det kan orsaka personskador på
grund av det kontinuerliga manschettrycket.
Enheten och manschetten är inte vattentäta. Undvik regn, svett ochvatten på enheten och manschetten.
Mätningarna kan bli felaktiga om enheten används i närheten av
TV-apparater, mikrovågsugnar, mobiltelefoner eller röntgenutrustningeller andra enheter med kraftiga magnetfält.
Svenska 2Trådlösa kommunikationsenheter, t.ex. enheter i hemmanätverk,
mobiltelefoner, trådlösa telefoner och deras basstationer samt
walkie-talkies kan påverka blodtrycksmätaren. Därför ska sådana enheterhållas på minst 30 cm avstånd.
Kontrollera att enheten är ren när den återanvänds.
Använd utrustning, delar och batterier betraktas inte som vanligthushållsavfall och måste kasseras i enlighet med lokala föreskrifter.
När nätadaptern används, kontrollera att nätadaptern enkelt kan dras ur
eluttaget vid behov.
Gör inga förändringar på enheten. Det kan orsaka olyckor eller skada
enheten.
För att mäta blodtrycket måste armen tryckas ihop av manschetten
tillräckligt hårt för att blodflödet genom artären ska stoppas tillfälligt. Det
kan orsaka smärta, domningar eller ett tillfälligt rött märke på armen. Detta
tillstånd uppträder särskilt vid upprepade på varandra följande mätningar.
Eventuella smärtor, domningar eller röda märken försvinner med tiden.
Om blodtrycket mäts för ofta kan det leda till skador på grund av störningar i
blodflödet. Kontrollera att användningen av enheten inte leder till långvarig
försämring av blodcirkulationen när enheten används upprepade gånger.
Om du genomgått mastektomi ska du kontakta läkare innan du använder
enheten.
Låt inte barn använda enheten själva och använd inte enheten inom
räckhåll för småbarn. Det kan orsaka olyckor eller skador.
Det finns små delar som kan leda till risk för kvävning om små barnsväljer dem av misstag.
Koppla bort nätadaptern när den inte används under mätningen.
Användning av tillbehör som inte beskrivs i bruksanvisningen kanäventyra säkerheten.
Om batteriet kortsluts kan det bli varmt och möjligen orsaka brännskador.
Låt enheten anpassa sig till omgivningen innan den används (ungefär entimme).
Kliniska prövningar har inte genomförts på nyfödda barn och gravida
kvinnor. Får inte användas på nyfödda barn eller gravida kvinnor.
Vidrör inte batterierna, likströmsuttaget och patienten samtidigt. Det kan
leda till elektriska stötar.
Blås inte upp manschetten om den inte sitter runt överarmen.

Kontraindikationer

Följande är försiktighetsåtgärder för korrekt användning av enheten.
Sätt inte manschetten på en arm som är ansluten till annan medicinsk
elektrisk utrustning. Utrustningen fungerar eventuellt inte korrekt.
Personer med allvarlig cirkulationsrubbning i armen måste kontakta en
läkare före användning av enheten för att undvika medicinska problem.
Svenska 3

Ställ ingen diagnos själv baserat på mätresultaten och starta ingen
behandling själv. Kontakta alltid läkare för en bedömning av resultaten
och behandlingen.
Sätt inte manschetten på en arm med ett sår som inte läkt.
Sätt inte manschetten på en arm som används för intravenöst dropp eller
blodtransfusion. Det kan orsaka skador eller olyckor.
Använd inte enheten i närheten av lättantändliga gaser som narkosgaser.
Det kan orsaka explosion.
Använd inte enheten i omgivningar med hög koncentration av syrgas, t.ex.
syrgaskammare med högtryck eller syrgastält. Det kan orsaka brand eller
explosion.

Svenska 4Ingående delar
Skärm

Luftkoppling

Kontakt för nätadapter
Likströmsuttag

START -knapp

Luftanslutning

LuftslangOmfång för korrekt
passform

Positioneringsmärke för artär

Indexmärke
ArmmanschettBatterifack

Batterilucka

1,5 V batterier(R6P, LR6 eller AA)

Skärm

MINNE

Genomsnitt

Symbol för IHB/AFib

Systoliskt
tryck%IHB/AFib

Symbol för fel manschettstorlek

Diastoliskt
tryck

Symbol för rörelsefel
Hjärtsymbol
Indikator för WHO-klassificeringoch tryckindikator

Pulsfrekvens

Batteriindikator
Svenska 5

Symboler
Symboler på enhetens hölje
Symboler
SYS
DIA
PUL

IP

Funktion/innebörd
Standby och slå på enheten.
Systoliskt blodtryck i mmHg
Diastoliskt blodtryck i mmHg
Puls per minut
Guide för installation av batteriet
Likström
Typ BF: Enheten, manschetten och slangen är utformade så att
de ger särskilt skydd mot elektriska stötar.
Märkning för medicinteknisk produkt enligt EU-direktivet
EU-representant
Tillverkare
Tillverkningsdatum
Symbol för internationell skyddsklass
WEEE-märkning

SN

Serienummer
Se bruksanvisningen/broschyren
Likströmsuttagets polaritet
Förvaras torrt

Symboler som visas på skärmen
Symboler

Funktion/innebörd
Visas när mätning pågår. Blinkar när
pulsen identifieras.
Symbol för IHB/AFib
Visas när hjärtrytmrubbningar
identifieras.
Kan tändas när en mycket liten vibration
som en rysning eller skakning
identifieras.
Visas när arm- eller kroppsrörelse
identifieras.
Svenska 6

RekommenderadåtgärdMätning pågår. Varså stilla som möjligt.

Mätningen kan ge ett
felaktigt värde.
Mät igen. Var stillaunder mätningen.

Symboler
Symboler

Funktion/innebörd
Visas under mätningen när
manschetten sitter löst
Identifierad frekvens för IHB/AFib iminnet%IHB/AFib =Antal identifierade
IHB/AFibs i minnet
Totalt antal

Rekommenderad åtgärd
Mätningen kan ge ett
felaktigt värde. Sätt på
manschetten korrekt och
mät igen.

x100 [%]

Tidigare mätningar sparade i
MINNET.
Genomsnittsvärde
LADDAT BATTERIBatteriströmindikatorn undermätning.
LÅGT BATTERIBatteriet är svagt när den blinkar.

Ojämnt blodtryck på grund av
rörelser under mätningen.
De systoliska och diastoliska
värdena ligger inom 10 mmHg frånvarandra.
Tryckvärdet ökade inte under
uppblåsning.
Manschetten sitter inte korrekt.
PUL-FEL
Pulsen identifierades inte korrekt.
Internt fel i blodtrycksmätaren

Svenska 7

Byt ut alla batterier mot
nya när symbolen
blinkar.
Mät igen.
Var mycket stilla undermätningen.

Sätt på manschetten
korrekt och mät igen.

Ta ut batterierna, tryck på
START -knappen ochsätt in batterierna igen.
Kontakta återförsäljarenom felet kvarstår.

Driftsläge
1. Normal mätning

Tryck på START -knappen. Blodtrycket mäts och data sparas i minnet.
Den här enheten kan spara de senaste 60 mätningarna i minnet.

2. Hämta data

När ingenting visas, håll in START -knappen.

I standby-läget,håll inSTART

knappenSläpp knappen när genomsnittsvärden visas.

Datanumret och sparade värden visas
automatiskt sorterade från den senaste
mätningen.

Information om hur man hämtar värden finns i
avsnitt “Hämta minnesdata“.

3. Radera alla data som sparats i minnetHåll in START -knappen tills “stänga av enheten.
“ för att radera data.
Välj “blinkar.
Data tas bort när
Enheten stängs av automatiskt.

“ visas för att

4. Mäta med önskat systoliskt tryck

På sida 15 visas hur man mäter med ett önskat systoliskt tryck.

Svenska 8

START

Använda mätaren
Installera/byta batterier

1. Ta bort batteriluckan.
2. Ta ut de förbrukade batterierna ochsätt in nya batterier i batterifacket
som på bilden. Var noga med att
polerna (+ och -) sitter korrekt.
Använd endast batterier av typen
R6P, LR6 eller AA.
3. Montera batteriluckan.

Steg 1
Steg 3

Steg 1
Steg 2

AKTA
Sätt in batterierna i batterifacket som på bilden. Om de sätts in på fel sättfungerar inte enheten.
Byt alla batterier mot nya när(LÅGT BATTERI-indikator) blinkar på
skärmen. Blanda inte gamla och nya batterier. Det kan förkorta batteriernas
livslängd eller göra så att enheten fungerar på fel sätt.
Vänta i två sekunder eller mer efter att enheten stängts av innan du byter
batterierna.
Utför en blodtrycksmätning om(LÅGT BATTERI-indikator) visas ävenefter att batterierna har bytts. Enheten kan då känna igen de nya batterierna.
(LÅGT BATTERI-indikator) visas inte när batterierna är slut.
Batteriernas livslängd varierar med omgivande temperatur och kan vara
kortare vid låga temperaturer. I allmänhet räcker fyra nya LR6-batterier
ungefär ett år vid mätning två gånger per dag.
Använd endast angivna batterityper. Batterierna som medföljer enheten ärtill för att testa mätarens funktion och kan ha en begränsad livstid.
Ta ut batterierna om enheten inte ska användas under en längre tid.
Batterierna kan läcka och orsaka felfunktion.

Svenska 9

Använda mätaren
Ansluta luftslangen

Luftanslutning

Sätt in luftkopplingen korrekt iluftanslutningen.

Luftkoppling

Ansluta nätadapternSätt i kontakten för nätadaptern ilikströmsuttaget.
Anslut sedan nätadaptern till ett
eluttag.
Använd den specificerade nätadaptern.
(Se sida 22.)

Likströmsuttag

Kontakt för nätadapterHåll i nätadapterns hus och dra utåt när nätadaptern ska kopplas bortfrån eluttaget.
Håll i kontakten för nätadaptern och dra utåt när nätadaptern ska
kopplas bort från blodtrycksmätaren.

Svenska 10

Använda mätaren
Välja rätt manschettstorlek

Det är viktigt att rätt manschettstorlek används för en korrekt mätning. Om
manschetten inte har rätt storlek kan mätningen ge ett felaktigt
blodtrycksvärde.
Armstorleken är angiven på manschetten.
Indexetoch omfång för korrekt passform på manschetten, visar om duanvänder korrekt manschett. (Se “Symboler på manschetten” på nästa sida.)
Om indexet
pekar utanför omfånget, kontakta den lokala återförsäljaren
för att köpa en ny manschett.
Armmanschetten är en förbrukningsvara. Köp en ny om den blir sliten.

Rekommenderad
Katalognummer
manschettstorlek
31 till 45 cm
Stor vuxenmanschett
CUF-F-LA
22 till 42 cm
Manschett med stort omfång
CUF-I
22 till 32 cm
Vuxenmanschett
CUF-F-A
Armstorlek: Omkretsen vid biceps.
Obs!
UA-767S är inte avsedd att användas med en liten manschett.
Armstorlek

Sätta på armmanschetten

1. Lägg manschetten runt överarmen,
ungefär 1–2 cm över armbågens
insida som på bilden.
Placera manschetten direkt mot
huden eftersom klädesplagg kan
göra att pulsen blir svag och leda till
felaktig mätning.

Rulla inte upp ärmen hårt.
Manschett
1–2 cmLuftslang
Positioneringsmärke för artär

2. Åtdragning runt överarmen på
grund av att en skjortärm rullats
upp kan ge felaktiga värden.

Fäst tygfästet
ordentligt.

3. Bekräfta att indexet
pekar inomomfånget för korrekt passform.
Index

Svenska 11

och omfång för korrekt passformAnvända mätaren
Symboler på manschetten
Symboler Funktion/innebörd
Positioneringsmärke för artär
▲
REF

Index
KatalognummerOmfång för korrekt passform för
vuxenmanschetten.
Angivet på vuxenmanschetten.

Rekommenderad åtgärd
Placera märket
påöverarmens artär eller ilinje med ringfingret påarmens insida.

AnvänddenstoraÖver omfånget som är angivet på vuxenmanschetten istället
vuxenmanschetten och manschetten för
vuxenmanschetten
med stort omfång.
eller manschetten med
stort omfång.
Omfång för korrekt passform för
manschetten med stort omfång.
Angivet på manschetten med stortomfång.
Omfång för korrekt passform för den
stora vuxenmanschetten. Angivet påden stora vuxenmanschetten.
Under omfånget som är angivet på
vuxenmanschetten och manschettmed stort omfång.
Använd vuxenmanschetten
Under omfånget på den stora istället för den stora
vuxenmanschetten.
vuxenmanschetten.
Lotnummer
Stor vuxenmanschettAOmfång för korrekt passform

L

Manschett med stort omfång

S

W

L

A

L

VuxenmanschettS

Svenska 12

Använda mätarenSå här görs korrekta mätningar

För noggranna blodtrycksmätningar:
Sitt bekvämt på en stol. Vila armen på bordet. Korsa inte benen. Hållfötterna platt mot golvet och räta på ryggen.
Koppla av i ungefär fem till tio minuter före mätningen.
Placera manschettens mitt i höjd med hjärtat.
Var stilla och tyst under mätningen.
Mät inte direkt efter fysisk aktivitet eller bad. Vila i tjugo eller trettio minuterföre mätningen.
Försök att mäta blodtrycket vid samma tidpunkt varje dag.

Mätning

Under mätningen är det normalt att manschetten känns väldigt trång. (Bliinte orolig.)

Efter mätningen

Tryck på START -knappen för att stänga av enheten efter mätningen.
Ta bort manschetten och anteckna dina värden.

Obs! Enheten har en automatisk avstängningsfunktion som stänger av
enheten ungefär en minut efter mätningen.
Vänta minst tre minuter mellan mätningar på samma person.

Svenska 13

Mätningar
Före mätning, se avsnitt “Anmärkningar för korrekt mätning“ på nästa
sida före mätning.

Normal mätning
1.

I höjd med hjärtat
Placera manschetten på armen (helst vänster arm).
Var tyst under mätningen.

Tryck

2.

Tryck på START -knappen.
Alla displaysegment visas.
Alla skärmens
Noll (0) blinkar under en kort tid.
segment visas
Skärmen ändras, som visas i figuren till
höger, när mätningen börjar. Manschettenbörjar blåsas upp. Det är normalt att
manschetten känns väldigt trång. En
tryckindikator visas till vänster på
Visar nollskärmen under uppblåsningen.
Startar uppblåsning
Obs! Tryck på START -knappen på nytt
om du vill avbryta uppblåsningen.

3.

När uppblåsningen är klar startas
tömningenautomatisktoch(hjärtsymbolen) blinkar, vilket indikerar att
mätning pågår. När pulsen identifieras
blinkar symbolen vid varje pulsslag.
Obs! Om ett korrekt tryck inte erhålls startar
enheten uppblåsningen igen automatiskt.
För att undvika ny uppblåsning, se avsnitt“Mäta med önskat systoliskt tryck“ pånästa sida.

4.

När mätningen är klar visas det systoliska
och diastoliska trycket samt pulsfrekvensen.
Manschetten släpper ut kvarvarande luft
och töms helt.

5.

Tryck på START -knappen för att stänga
av enheten.

START

Trycksätter

Mätningpågår

Systoliskt tryck
Diastoliskt tryck
WHO-klassificering

Pulsfrekvens
Obs! Vänta minst tre minuter mellan
mätningar på samma person.

Svenska 14

Pressar ut kvarvarande
luft automatisktMätningar
Mäta med önskat systoliskt tryck

UA-767S är utformad så att den identifierar pulsen och blåser upp
manschetten till ett systoliskt tryck automatiskt.
Använd den här metoden när manschetten blåses upp på nytt upprepadegånger eller när resultaten inte visas även om trycket minskar till
20 mmHg eller lägre.

1.

Placera manschetten på armen i höjd
med hjärtat (helst vänster arm).

2.

Tryck på START -knappen.

3.

När nollan blinkar, håll in START-knappen tills ett värde som är cirka
30 till 40 mmHg högre än dittförväntade systoliska tryck visas.

4.

Släpp START -knappen för att starta
mätningen när önskat värde visas.
Fortsätt att mäta blodtrycket enligt
beskrivningen på föregående sida.

Anmärkningar för korrekt mätning

I höjd med hjärtat
Tryck på START
knappenSläpp knappen vid
det önskade
systoliska trycketHåll inknappen

START

Se föregående sida för
mätning

Sitt i en bekväm position. Lägg armen på ett bord med handflatan uppåt
och manschetten i höjd med hjärtat.
Koppla av i ungefär fem till tio minuter före mätningen. Om du är
upphetsad eller nedstämd av emotionell stress visar mätningen den
stressen som ett högre (eller lägre) blodtrycksvärde än normalt och
pulsvärdet kommer vanligtvis att vara högre än normalt.
En människas blodtryck varierar kontinuerligt beroende på vad man gör
och vad man har ätit. Vad man dricker kan ha en mycket stor och snabb
effekt på blodtrycket.
Enheten baserar mätvärdena på hjärtslagen. Om du har mycket svaga
hjärtslag eller hjärtrytmrubbningar kan enheten att ha svårt att bestämma
ditt blodtryck.
Om enheten identifierar ett onormalt tillstånd avbryts mätningen och en
felsymbol visas. Se sida 7 för en beskrivning av symbolerna.
Svenska 15

Blodtrycksmätaren är avsedd att användas av vuxna. Kontakta läkare
innan enheten används på barn. Barn ska inte använda enheten utanövervakning.
Den automatiska blodtrycksmätarens funktion kan påverkas av för hög
temperatur, luftfuktighet eller altitud.

Hämta minnesdata
Obs! Enheten kan spara de senaste 60 mätningarna i minnet.
1. Håll in _START_ -knappen för att
hämta sparade värden, när
ingenting visas.

Tryck och START knappenhåll in
Genomsnittligt systoliskt
Genomsnittligt diastoliskt
Genomsnittlig puls

2. Släpp knappen när
genomsnittsvärden visas.

Genomsnitt för
alla data

3. Datanumret och sparade värden
visas automatiskt sorterade från den
sista mätningen.
Systoliskt
4. Skärmen stängs av automatiskt efter
att alla värden visats.

Diastoliskt
Puls
Senaste data

Obs! Om du trycker på_START_ -knappen medan
värden hämtas stängs enheten
av.

Systoliskt
Diastoliskt
Puls
Sista data (äldsta)

Svenska 16

Vad är IHB/AFib-indikatorn?
När mätaren identifierar en oregelbunden frekvens under mätningenvisas IHB/AFib-indikatorn på skärmen med mätvärdena.
Obs! Vi rekommenderar att du
IHB/AFib-indikatorn ofta visas.

kontaktar

läkareom

Vad är AFib (förmaksflimmer)?
Hjärtat drar ihop sig på grund av elektriska signaler i hjärtat och skickar ut
blod genom kroppen. Förmaksflimmer (AFib) inträffar när den elektriska
signalen i förmaket störs och leder till störningar i pulsintervallet.
Förmaksflimmer (AFib) kan leda till att blod stannar i hjärtat och kan lättbilda koagel, vilket är en orsak till stroke och hjärtattack.

%IHB/AFib (arytmi/förmaksflimmer)%IHB/AFib (arytmi/förmaksflimmer) visas som frekvensen för att
identifiera IHB/AFib (arytmier/förmaksflimmer).
IHB/AFib kan inte bara identifiera brus som fysisk rörelse utan ävenhjärtrytmrubbningar. Vi rekommenderar därför att du kontaktar läkare
om %IHB/AFib är högt.
Antal identifierade
IHB/AFibs i minnet
x 100 [%]
%IHB/AFib =Totalt antal
Visning av %IHB/AFib: %IHB/AFib visas när genomsnittliga värden visas.
(Se avsnitt “2. Hämta värden“ i “Driftsläge“)%IHB/AFib (arytmi/förmaksflimmer) visas inte när
antalet i minnet är sex eller lägre.
Visning av genomsnittliga värden

Nivå 0%IHB/AFib=0Nivå 1Nivå 2Nivå 3%IHB/AFib=1–9 %IHB/AFib=10–24 %IHB/AFib=25–100

Visas inte

Svenska 17

Tryckindikator
Indikatorn visar tryckets förlopp under mätningen.
Trycksätter

Släpper ut luft

Uppblåsning pågår

Mätning pågår
Uppblåsning klar

Indikator för WHO-klassificering
Varje segment i indikatorn motsvarar
WHO:sblodtrycksklassificering
enligtbeskrivningen på nästa sida.

Exempel:Indikator för WHO-klassificering
Allvarlig hypertoniMåttlig hypertoni
Lindrig hypertoni
Hög normal
NormalOptimal: Indikatorn visar ett segment, baserat på
aktuella data, som motsvarar
WHO-klassificeringen.

Måttlig hypertoni

Lindrig hypertoni

Hög normal

Om blodtryck
Vad är blodtryck?

Blodtryck är den kraft som hjärtat utövar mot artärväggarna. Systoliskt
tryck är blodtrycket vid hjärtats sammandragning. Diastoliskt tryck är
blodtrycket vid hjärtats utvidgning. Blodtrycket mäts i millimeter
kvicksilver (mmHg). En människas naturliga blodtryck representeras av
det grundläggande trycket som mäts direkt på morgonen när man är
utvilad och innan måltid.
Svenska 18

Var är hypertoni och hur kontrolleras det?

Hypertoni – ett onormalt högt artärtryck – kan, om det inte behandlas,orsaka flera hälsoproblem inklusive stroke och hjärtattack. Hypertoni kan
kontrolleras genom att ändra livsstil, undvika stress och med läkemedel
under läkares övervakning.
För att förhindra hypertoni eller hålla det under kontroll:
Rök inte
Motionera regelbundet
Minska intaget av salt och fett
Gör regelbundna hälsokontrollerHåll en lämplig vikt

Varför mäta blodtrycket i hemmet?

När blodtrycket mäts på sjukhus eller läkarmottagning kan det skapaängslan som kan ge ett förhöjt värde, 25 till 30 mmHg högre än det som
mäts i hemmet. Mätning i hemmet minskar effekterna av yttre påverkanpå blodtrycksvärdena, utgör ett komplement till läkarens mätningar och
ger en mer exakt och fullständig blodtryckshistorik.

WHO:s blodtrycksklassificering

Variationer i blodtrycket

Diastoliskt blodtryck

Standarder för bedömning av högt blodtryck, oavsett ålder, har upprättats
Referensmaterial: Journal of Hypertension 1999, Vol 17 No. 2
av
mmHg
Hypertoni grad 3 (allvarlig)
Världshälsoorganisationen
110(WHO), som visas i
Hypertoni grad 2 (måttlig)
105
diagrammet.
100
95Hypertoni grad 1 (lindrig)

90

Blodtryck

Hög-normal
En människas blodtryck
85
Normalvarierar mycket dag till dag
80
Optimaloch beroende på årstid. Det
120 130 140 150 160 170 180
kan variera med 30 till
Systoliskt blodtryck
mmHg
50 mmHg på grund av olika
förutsättningar under dagen. Hos människor med hypertoni ärvariationerna ännu större. Normalt stiger blodtrycket vid arbete eller lek
och sjunker till sin lägsta nivå när man sover. Bli därför inte alltför oroadöver resultaten från en mätning.
Mät vid samma tid varje dag med metoden
Typiska fluktuationer under en dag
som beskrivs i bruksanvisningen så att du(mätt var femte minut)
lär känna ditt normala blodtryck.
Systoliskt
Regelbundna mätningar ger en mer
fullständig blodtryckshistorik. Var noga
med att notera datum och tid när du mäter
ditt blodtryck. Kontakta läkare för att tolka
Diastoliskt
Sömn
dina blodtrycksvärden.
PM

Svenska 19

AM

Tid

Felsökning
Problem
Ingenting visas
på skärmen,även när
enheten ärpåslagen.

Möjlig orsak
Batterierna är slut.
Batteriernas poler ärinte i rätt position.

Batterispänningen är för
låg.
(LÅGT
Manschetten
BATTERI-indikator)blåses inte upp.
blinkar. Om batterierna
är helt urladdade visas
inte symbolen.
Manschetten sitter intekorrekt.
Du rörde armen eller
kroppen undermätningen.
Enheten mäter
inte.
Manschettens position
Värdena är för
är inte korrekt.
höga eller för
låga.

Rekommenderad åtgärd
Byt ut alla batterierna mot nya.
Sätt in batterierna så att den
negativa och den positiva polen
matchar de som visas ibatterifacket.

Byt ut alla batterierna mot nya.

Sätt på manschetten korrekt.
Var helt stilla och tyst undermätningen.
Sitt bekvämt och stilla.
Lägg armen på ett bord med
handflatan uppåt och
manschetten i höjd med hjärtat.
Om du har mycket svaga
hjärtslag eller
hjärtrytmrubbningar kan
enheten att ha svårt att
bestämma ditt blodtryck.

Värdet skiljer sig fråndet som uppmätts på en Se avsnitt “Varför mäta
klinik eller
blodtrycket i hemmet?“.
Övrigtläkarmottagning.
Ta ut batterierna. Sätt tillbaka
dem korrekt och mät igen.
Obs! Kontakta återförsäljaren om åtgärderna som beskrivs ovan inte löser
problemet. Försök inte att öppna eller reparera produkten – garantin
gäller då inte längre.

Svenska 20

Underhåll
Öppna inte enheten. Den har ömtåliga elektriska komponenter och en
invecklad luftenhet som kan skadas. Om problemet inte kan lösas med
felsökningsanvisningarna, kontakta den auktoriserade återförsäljaren i dittområde eller vår kundservice. Kundservice hos A&D ger teknisk information,
tillhandahåller reservdelar och enheter till auktoriserade återförsäljare.
Enheten har utformats och tillverkats för att få en lång livslängd. Vi
rekommenderar emellertid normalt att låta inspektera enheten vartannat
år för att säkerställa korrekt funktion och noggrannhet. Kontakta den
auktoriserade återförsäljaren i ditt område eller A&D för underhåll.

Tekniska data
Typ
Mätmetod
Mätintervall

Mätnoggrannhet
Strömförsörjning
Antal mätningar

Klassificering

Kliniskt test
EMD
Minne
Driftförhållanden
Transport-/förvaringsförhållanden
Mått
Vikt

UA-767S
Oscillometrisk mätning
Tryck:
0–299 mmHg
Systoliskt tryck: 60–279 mmHg
Diastoliskt tryck: 40–200 mmHg
Puls:
40–180 slag/minut
Tryck:±3 mmHgPuls:±5 %4 x V 1,5 batterier (R6P, LR6 eller AA) eller
Nätadapter (tillval) (TB-233C) (medföljer inte)
Cirka 700 gånger
LR6 (alkaliska batterier)Cirka 200 gånger
R6P (manganbatterier)
Med ett tryckvärde på 180 mmHg vid
rumstemperatur (23 °C).
Internt strömförsörjd medicinsk elektrisk utrustning(drivs med batterier)/klass II (drivs med
nätadapter)
Kontinuerligt driftläge
I enlighet med ISO 81060-2: 2013
I den kliniska valideringsstudien användes K5 på 85
subjekt för bestämning av diastoliskt blodtryck.
IEC 60601-1-2: 2014
Senaste 60 mätningarna+10 till +40 °C/15 till 85 % RH/800 till 1 060 hPa-20 till +60 °C/10 till 95 % RH/700 till 1 060 hPa
Ca 140 [B] x 60 [H] x 105 [D] mm
Ca 245 g, exklusive batterier
Svenska 21

Kapslingsklass

Enhet:

IP20

Applicerad delAnvändbar livstid

Manschett
Enhet:Typ BF5 år (vid användning sex gånger per
dag)
Manschett: 2 år (vid användning sex gånger per
dag)
Nätadapter: 5 år (vid användning sex gånger per
dag)

Nätadapter som tillbehör
Adaptern är avsedd för att ansluta enheten till en strömkälla i hemmet.
Kontakta den lokala A&D-återförsäljaren för inköp. Nätadaptern måste
inspekteras eller bytas ut regelbundet.
TB-233C
Kontakta lokala A&D-återförsäljaren för inköp.
Nätadaptern måste inspekteras eller bytas ut
regelbundet.
Symboler på nätadaptern
Symboler
Funktion/innebörd
Endast för inomhusbruk
Enhet av klass II
Termisk säkring
Säkring
Märkning för medicinteknisk produkt
enligt EU-direktivet
Märkning för certifiering enligt EAC
Polaritet för kontakten för nätadaptern
Tillbehör säljs separat
Manschett Katalognummer
Manschettstorlek
CUF-F-LA
Stor vuxenmanschett
CUF-I
Manschett med stort omfång
CUF-F-A
VuxenmanschettArmstorlek: Omkretsen vid biceps.
Nätadapter
Obs!

Katalognummer
TB-233C

Armstorlek
31 till 45 cm
22 till 42 cm
22 till 32 cm

Kontakt (för eluttag)
Typ CSpecifikationer kan ändras utan föregående meddelande.
IP-klassificeringen är den nivå av skydd som kapslingen ger enligt IEC
60529. Enheten är skyddad mot fasta främmande föremål med 12 mm
diameter och större, t.ex. ett finger. Enheten är inte skyddad mot vatten.

Svenska 22

InnholdKjære kunder ........................................................................ 2
Innledende merknader .......................................................... 2
Forholdsregler ...................................................................... 2
Identifisering av deler ........................................................... 5
Symboler .............................................................................. 6
Driftsmodus ......................................................................... 8
Bruke måleren ...................................................................... 9Sette inn/skifte batterier .................................................. 9
Koble til luftslangen ....................................................... 10
Koble til strømadapteren ............................................... 10
Velge riktig mansjettstørrelse ........................................ 11
Sette på mansjetten ....................................................... 11
Foreta nøyaktige målinger ............................................. 13Måling ........................................................................... 13
Etter målingen ............................................................... 13Målinger ............................................................................. 14
Normal måling ............................................................... 14Målinger med ønsket systolisk trykk .............................. 15
Merknader for nøyaktig måling ...................................... 15
Hente frem data fra minnet ................................................. 16
Hva er IHB/AFib-indikatoren?.............................................. 17
Hva er AFib?........................................................................ 17％IHB/AFib .......................................................................... 17
Trykkstolpeindikator ........................................................... 18
WHO-klassifiseringsindikator.............................................. 18
Om blodtrykk ..................................................................... 18
Hva er blodtrykk? ........................................................... 18
Hva er hypertensjon, og hvordan kan det kontrolleres? ..... 19
Hvorfor måle blodtrykket hjemme? ................................ 19
WHOs klassifisering av blodtrykk ................................... 19
Variasjoner i blodtrykket ................................................ 19
Feilsøking ........................................................................... 20
Vedlikehold ........................................................................ 21
Tekniske spesifikasjoner ..................................................... 21

Norsk 1Kjære kunder
Gratulerer med anskaffelsen av en førsteklasses A&D blodtrykkmåler.
Enheten er utviklet for å være et brukervennlig og nøyaktig hjelpemiddel iden daglige overvåkingen av blodtrykket.
Vi anbefaler å lese denne håndboken nøye før du bruker enheten for
første gang.

Innledende merknader
Denne enheten samsvarer med EU-direktiv 93/42 EEC for medisinsk
bekrefter
dette.
(0123:
utstyr.
Samsvarsmerket
Referansenummeret til det involverte tekniske kontrollorganet)
Enheten er designet for bruk på voksne, ikke på spedbarn eller barn.
Bruksmiljø. Enheten er ment for å betjenes av brukeren i ethjemmepleiemiljø.
Denne enheten er designet for å måle blodtrykk og puls på mennesker for
diagnoseformål.

Forholdsregler
Enheten er konstruert med presisjonskomponenter. Unngå ekstreme
temperaturer, fuktighet, direkte sollys, slag og støv.
Rengjør enheten og mansjetten med en tørr, myk klut eller en klut fuktet
med vann og et nøytralt vaskemiddel. Bruk aldri alkohol, benzen, tynnereller andre sterke kjemikalier til å rengjøre enheten eller mansjetten.
Unngå å brette mansjetten stramt og å lagre slangen stramt oppkveilet i
lange perioder. Slik behandling kan forkorte komponentenes levetid.
Oppbevares utenfor rekkevidde for små barn, da slangen og kabelen kan
utgjøre en kvelningsfare.
Unngå å vri luftslangen under måling. Det kan forårsake personskade
som følge av kontinuerlig mansjettrykk.
Verken enheten eller mansjetten er vanntett. Unngå at enheten og
mansjetten utsettes for regn, svette eller vann.
Målingen kan bli forstyrret hvis enheten brukes nær TV-apparater,
mikrobølgeovner, mobiltelefoner, røntgenutstyr eller annet utstyr med
sterke elektriske felt.
Trådløse kommunikasjonsenheter som hjemmenettverkssendere,mobiltelefoner, trådløse telefoner og basestasjoner, samt walkie-talkier
kan påvirke denne blodtrykksmåleren. Det anbefales derfor å sørge foren avstand på minst 30 cm til slike apparater.
Norsk 2

Kontroller at enheten er ren før den brukes igjen.
Brukt utstyr, deler og batterier skal ikke behandles som vanlig
husholdningsavfall, og må kasseres i samsvar med gjeldendebestemmelser.
Sørg for at strømadapteren enkelt kan kobles fra stikkontakten om
nødvendig når apparatet brukes med strømadapteren.
Ikke foreta inngrep i enheten. Det kan forårsake ulykker eller skade på
enheten.
For å måle blodtrykket må mansjetten klemme hardt nok over armen til at
blodstrømmen gjennom arterien stoppes midlertidig. Det kan føre til
smerte, nummenhet eller et midlertidig, rødt merke på armen. Dette
forekommer særlig hvis det foretas flere målinger etter hverandre.
Eventuell smerte, nummenhet eller rødhet merker forsvinner etter enstund.
For hyppig måling av blodtrykket kan forårsake skade som følge av atblodsirkulasjonen hindres. Pass på at bruken av enheten ikke fører til
langvarig hindring av blodsirkulasjonen hvis enheten brukes flere ganger
etter hverandre.
Hvis du har fått utført mastektomi, må du rådføre deg med lege før dubruker denne enheten.
Ikke la barn bruke enheten alene, og unngå å bruke enheten innenfor
rekkevidde for barn. Det kan forårsake ulykker eller skade.
Enheten inneholder små deler som kan medføre kvelningsfare hvis deved et uhell svelges av et barn.
Trekk strømadapteren ut av stikkontakten når den ikke er i bruk undermåling.
Bruk av tilbehør som ikke er beskrevet i denne håndboken kan gå ut oversikkerheten.
Batteriet kan bli varmt nok til å forårsake forbrenning hvis det kortsluttes.
La enheten tilpasses til omgivelsene før den brukes (om lag en time).
Det er ikke utført klinisk testing på nyfødte spedbarn eller gravide kvinner.
Enheten må ikke brukes på nyfødte spedbarn eller gravide kvinner.
Ikke berør batteriene, likestrømspluggen og pasienten samtidig. Det kan
føre til elektrisk støt.
Mansjetten må ikke blåses opp når den ikke er festet rundt overarmen.

Kontraindikasjoner
Følgende punkter er forholdsregler for riktig bruk av enheten.
Mansjetten må ikke festes på armen hvis annet medisinsk elektrisk utstyr
er satt på. Det kan føre til at utstyret ikke fungerer som det skal.
Personer som har alvorlig nedsatt blodomløp i armen, må rådføre seg
med lege før enheten brukes, for å unngå medisinske problemer.

Norsk 3

Du må ikke stille diagnose på deg selv eller starte behandling på grunnlag
av måleresultatene. Rådfør deg alltid med en lege for å få evaluert
resultatene og behandlingen.
Mansjetten må ikke settes på en arm med et åpent sår.
Mansjetten må ikke settes på en arm som brukes til intravenøst drypp
eller blodoverføring. Det kan forårsake personskade eller ulykker.
Ikke bruk enheten på steder der det finnes brannfarlig gass, for eksempel
anestesigass. Det kan forårsake eksplosjon.
Ikke bruk enheten i oksygenanrikede miljøer, for eksempel
oksygentrykkammer eller oksygentelt. Det kan forårsake brann eller
eksplosjon.

Norsk 4

Identifisering av deler
Display

Plugg forvekselstrømadapter

Luftplugg

Likestrømsplugg
START -knappen

LuftkoblingOmråde for riktig
tilpasning

Luftslange
Arterieposisjonsmerke

Indeksmerke
Mansjett
Batterirom
Batterideksel

1,5 V batterier(R6P, LR6 eller AA)

Display

Gjennomsnitt

MINNE
IHB/AFib-symbol
Systolisk
trykk%IHB/AFib
Symbol for feil mansjettmontering

Diastolisk
trykk

Symbol for bevegelsesfeil

Hjertesymbol
Puls

WHO-klassifiseringsindikator
og trykkstolpeindikator

Batteriindikator
Norsk 5

Symboler
Trykte symboler på enhetens etui
Symboler
SYS
DIA
PUL

IP

Funksjon/betydningStandby og slå på enheten.
Systolisk blodtrykk i mmHg
Diastolisk blodtrykk i mmHg
Puls per minutt
Veiledning for batteriinstallasjon
Likestrøm
Type BF: Enheten, mansjetten og slangen er konstruert for å gi
spesiell beskyttelse mot elektrisk støt.
Symbol for EC-direktivet for medisinsk utstyr
EU-representant
Produsent
Produksjonsdato
Internasjonalt beskyttelsessymbol
WEEE-merke

SN

Serienummer
Se bruksanvisningen/veiledningsheftet
Likestrømspluggens polaritet
Holdes tørt

Symboler som vises på displayet
SymbolerFunksjon/betydning
Vises mens måling pågår. Blinker når
puls er registrert.
IHB/AFib-symbol
Vises hvis uregelmessig hjerterytme
registreres.
Kan tennes hvis enheten registrerersvært lett vibrasjon, for eksempel ved
skjelving eller risting.
Vises hvis enheten registrerer at
kroppen eller armen har bevegd seg.

Norsk 6

Anbefalt tiltak
Målingen pågår. Sitt såstille som mulig.

Måleren kan vise feil
verdi.
Foreta en ny måling.
Sitt stille mensmålingen pågår.

Symboler
SymbolerFunksjon/betydningAnbefalt tiltak
Måleren kan vise feil verdi.
Sett på mansjetten på
riktig måte, og foreta en nymåling.

Vises under måling hvis
mansjetten sitter løst
Registrert IHB/AFib-verdi i minnet%IHB/AFib =Antall registrerte
IHB/AFib-forekomster
i minnet

x100 [%]

Totalt antall

Tidligere målinger lagret i
MINNET.
Gjennomsnittsdata
FULLT BATTERIBatteristrømindikatoren undermåling.
LAVT BATTERINIVÅ
Symbolet blinker når batterinivået
er lavt.

Ustabilt blodtrykk på grunn av
bevegelse under måling.

Erstatt alle batteriene med
nye batterier når symbolet
blinker.
Foreta en ny måling.
Sitt helt stille mensmålingen pågår.

Forskjellen mellom systolisk og
diastolisk verdi er mindre enn
10 mmHg.
Trykkverdien økte ikke under
oppblåsing.
Mansjetten sitter ikke som den
skal.
PUL-DISPLAYFEIL
Pulsen ble ikke registrert på riktigmåte.

Sett på mansjetten på
riktig måte, og foreta en nymåling.

Intern feil i blodtrykkmåleren

Ta ut batteriene og trykk
på knappen START , og
sett deretter inn batteriene
igjen. Kontakt forhandleren
hvis feilen vedvarer.

Norsk 7

Driftsmodus
1. Normal måling
Trykk på knappen START . Blodtrykket måles, og informasjonen lagres iminnet. Enheten kan lagre de siste 60 målingene i minnet.

2. Hente frem data
Hvis ingenting vises, trykker du og holder inne
knappen START .

I standby,
trykk og hold inne
START

knappenSlipp knappen når gjennomsnittlige data vises.

Datanummeret og lagrede data vises
automatisk i rekkefølge fra den siste målingen.

Se flere detaljer om fremhenting av data i“Hente frem data fra minnet“.

3. Slette alle lagrede verdier i minnet
Slå av enheten ved å holde inne knappen
START til ““ vises.
Velg ““ for å slette verdiene.
Verdiene slettes når symbolet
blinker.
Enheten slås av automatisk.

4. Målinger med ønsket systolisk trykk
Se informasjon om måling med ønsket systolisk trykk på side 15.

Norsk 8START

Bruke måleren
Sette inn/skifte batterier
1. Fjern batteridekselet.
2. Fjern de brukte batteriene, og settinn nye batterier i batterirommet som
vist. Pass på riktig polaritet (+ og -).
Bruk bare R6P, LR6 eller
AA-batterier.
3. Sett på batteridekselet.

Trinn 1
Trinn 3

Trinn 1
Trinn 2FORSIKTIG
Sett batteriene inn i batterirommet som vist. Enheten fungerer ikke hvis
batteriene settes inn feil.
Når(symbol for LAVT BATTERINIVÅ) blinker på displayet, må allebatteriene erstattes med nye. Unngå å blande gamle og nye batterier. Det
kan forkorte batterienes levetid eller føre til at enheten ikke fungerer som
den skal.
Slå av enheten og vent minst to sekunder før batteriene skiftes.
Hvis symbolet(symbol for LAVT BATTERI) vises også etter atbatteriene er skiftet, foretar du en blodtrykksmåling. Det kan være atenheten registrerer de nye batteriene når målingen foretas.
(Symbolet for LAVT BATTERINIVÅ) vises ikke hvis batteriene er helt
utladet.
Batterilevetiden varierer avhengig av omgivelsestemperatur, og kan
forkortes i lave temperaturer. Fire nye LR6-batterier varer vanligvis ett år
ved måling to ganger hver dag.
Bruk bare de spesifiserte batteriene. Batteriene som følger med enheten er
ment for testing av målerens funksjon, og kan ha begrenset levetid.
Ta ut batteriene hvis enheten ikke skal brukes på en lang stund.
Batteriene kan lekke og føre til funksjonsfeil.

Norsk 9Bruke måleren
Koble til luftslangen

Luftkobling

Sett luftpluggen bestemt inn iluftkoblingen.

Luftplugg

Koble til strømadapteren
Sett vekselstrømadapterpluggen
inn i likestrømskontakten.
Koble deretter strømadapteren til
en stikkontakt.
Bruk den spesifiserte strømadapteren.
(Se side 22.)

Likestrømsplugg
Plugg forvekselstrømadapter

Hold i støpselet når strømadapteren trekkes ut av stikkontakten.
Hold i adapterpluggen
blodtrykkmåleren.

når

strømadapteren

Norsk 10trekkesut

av

Bruke måleren
Velge riktig mansjettstørrelse
Det er viktig å bruke riktig mansjettstørrelse for å oppnå riktige målinger.
Hvis mansjetten ikke har riktig størrelse, kan målingen vise feil
blodtrykksverdi.
Armstørrelsen er trykt på hver mansjett.
Indeksmerket
og festeområdet på mansjetten viser om du bruker riktig
mansjett. (Se “Trykte symboler på mansjetten” på neste side)
Hvis indeksmerket
er utenfor området, må du kontakte den lokale
forhandleren for å få kjøpt en ny mansjett.
Mansjetten er en forbruksartikkel. Kjøp en ny hvis den blir slitt.
Armstørrelse
Anbefalt mansjettstørrelse
Katalognummer
31 cm til 45 cm
Stor mansjett for voksne
CUF-F-LA
22 cm til 42 cm
Universalmansjett
CUF-I
22 cm til 32 cm
Mansjett for voksne
CUF-F-A
Armstørrelse: Armens omkrets ved biceps.
Merknad:
UA-767S er ikke ment for bruk med liten mansjett.

Sette på mansjetten
1. Legg mansjetten rundt overarmen,
om lag 1-2 cm over innsiden av
albuen, som vist.
Legg mansjetten rett på huden. Klærmellom mansjetten og huden kan
føre til svakt pulssignal og gi feilmåleresultat.

Unngå å brette opp ermet for stramt.
Mansjett
1-2 cmLuftslange
Arterieposisjonsmerke

2. Hvis blodomløpet i overarmen
hindres av at ermet brettes forstramt, kan det føre til unøyaktigemåleverdier.
3. Kontroller at indeksmerket
innenforområdetfortilpasning.

erriktig

Norsk 11

Fest tekstilfestet
godt.

Indeksmerkeog område for riktig tilpasning

Bruke måleren
Trykte symboler på mansjetten
Symboler Funksjon/betydningAnbefalt tiltak
påPlasser merket
arterien i overarmen ellerpå linje med ringfingerenpå innsiden av armen.

Arterieposisjonsmerke
Indeksmerke
KatalognummerOmråde for riktig tilpasningmansjett for voksne.
Trykt på mansjetten for voksne.

▲
REF

for

Bruk den store mansjetten
Over området trykt på mansjetten for for voksne i stedet forvoksne og universalmansjetten.
mansjetten for voksne eller
universalmansjetten.
Område for riktig tilpasning foruniversalmansjetten.
Tryktpå
universalmansjetten.
Område for riktig tilpasning for den
store mansjetten for voksne. Trykt på
den store mansjetten for voksne.
Under området som er trykt på
mansjetten
forvoksneog
universalmansjetten.
Bruk mansjetten for voksne
Under området på den store i stedet for den store
mansjetten for voksne.
mansjetten for voksne.
Lot-nummer
Stor mansjett for voksneOmråde for riktig tilpasning

A

L

Universalmansjett

S

W

L

A

L

Mansjett for voksneS

Norsk 12Bruke måleren
Foreta nøyaktige målinger
For mest mulig nøyaktige blodtrykksmålinger:
Sitt komfortabelt på en stol. Hvil armen på bordet. Ikke kryss bena. Holdføttene flatt på gulvet, og hold ryggen rett.
Slapp av i om lag fem til ti minutter før målingen.
Plasser midten av mansjetten på høyde med hjertet.
Sitt stille og unngå å snakke under målingen.
Unngå å måle like etter fysiske anstrengelser eller et bad. Hvil i 20 til 30
minutter før målingen foretas.
Prøv å måle blodtrykket på samme tidspunkt hver dag.

MålingDet er normalt at mansjetten føles svært stram mens målingen pågår. (Deter ingen grunn til bekymring.)

Etter målingen
Etter målingen trykker du på knappen START for å slå av enheten.
Fjern mansjetten og noter verdiene.
Merknad: Enheten slås av automatisk om lag ett minutt etter målingen.
Vent minst tre minutter mellom målinger på samme person.

Norsk 13Målinger
Se “Merknader for nøyaktig måling“ på neste side før målingen utføres.

Normal måling
1.

Sett mansjetten på armen (helst venstre arm).
På høyde med hjertet
Sitt stille mens målingen pågår.

2.

Trykk på knappen START .
Trykk påAlle displaysegmentene vises.
Null (0) vises, og blinker kort.
Displayet endres som vist i illustrasjonen
Alle displaytil høyre når målingen begynner.
segmentene
vises
Oppblåsing av mansjetten starter. Det er
normalt at mansjetten føles svært stram.
En trykkstolpeindikator vises på venstre
side av mansjetten mens oppblåsingen
Null displaypågår.
Starter oppblåsing
Merknad: Hvis du ønsker å stoppe
oppblåsingen på noe tidspunkt,kan du trykke på knappen
START igjen.

3.

4.

5.

START

Når oppblåsingen er fullført, slippes luft ut
Trykk bygges opp(hjertesymbolet)
automatisk, og
blinker for å indikere at målingen pågår.
Symbolet blinker med hvert hjerteslag såsnart pulsen er registrert.
Merknad: Hvis enheten ikke oppnår riktig trykk,
begynner enheten automatisk å blåse
opp mansjetten igjen.
Se “Målinger med ønsket systolisk
trykk“ på neste side for å unngå atmansjetten blåses opp på nytt.
MålingpågårNår målingen er fullført, vises systolisk og
diastolisk trykkverdi, samt puls.
Gjenværende luft slippes ut av mansjetten, Systolisk trykk
og den tømmes helt.
Diastolisk trykk
Trykk på knappen START for å slå av WHO-klassifisering
Puls
enheten.

Merknad: Vent minst tre minutter mellommålinger på samme person.
Norsk 14

Gjenværende luft
slippes ut automatiskMålingerMålinger med ønsket systolisk trykk
UA-767S er konstruert for å registrere puls og blåse opp mansjetten til
systolisk trykk automatisk.
Bruk denne metoden hvis mansjetten blåses opp igjen flere ganger ellerhvis resultatene ikke vises selv om trykket faller til 20 mmHg eller mindre.
1.

Sett mansjetten på armen på høyde med
hjertet (helst venstre arm).

2.

Trykk på knappen START .

3.

Mens tallet null blinker, trykker du påog holder inne knappen START til ettall som er om lag 30 til 40 mmHg
høyere enn forventet systolisk trykk
vises.

4.

Når ønsket verdi nås, slipper du Trykk på og hold START
inne knappen
knappen START for å starte målingen.
Fortsett å måle blodtrykket som
beskrevet på neste side.

På høyde med hjertet
Trykk påknappenSTART

Slipp knappen vedønsket systolisk
trykk

Se forrige side for målinger

Merknader for nøyaktig måling

Sitt i en komfortabel stilling. Legg armen på et bord med håndflaten vendt
opp og mansjetten på høyde med hjertet.
Slapp av i om lag fem til ti minutter før målingen. Hvis du er
følelsesmessig opphisset eller nedtrykt, vil målingen gjenspeile dettestresset som en høyere (eller lavere) blodtrykksverdi enn normalt, og
pulsverdien vil vanligvis være høyere.
En persons blodtrykk varierer hele tiden, avhengig av hva personen gjør
og hva personen har spist. Det du drikker kan ha sterk og rask virkning på
blodtrykket.
Enhetens målinger er basert på pulsen. Hvis du har svært svak elleruregelmessig puls, kan det være vanskelig for enheten å fastslå
blodtrykket ditt.
Hvis enheten registrerer en unormal tilstand, stoppes målingen og etfeilsymbol vises. Se beskrivelse av symbolene på side 7.
Denne blodtrykkmåleren er kun ment for å brukes av voksne. Rådfør deg
med legen din før enheten brukes på et barn. Barn skal ikke bruke
enheten uten tilsyn.
Norsk 15

Den automatiske blodtrykksmålerens funksjon kan bli påvirket av høy
temperatur, fuktighet eller høyde over havet.

Hente frem data fra minnet
Merknad: Enheten lagrer de siste 60 målingene i minnet.
1. Hvis ingenting vises, trykker du og
holder inne knappen _START_ forå hente frem lagrede data.

Trykk oghold inne

START

knappenGjennomsnittlig systolisk
Gjennomsnitt diastolisk

2. Slipp knappen når gjennomsnittlige
data vises.

Gjennomsnitt puls

Gjennomsnitt av
alle verdiene

3. Datanummeret og lagrede datavises automatisk i rekkefølge fra densiste målingen.
Systolisk
4. Displayet slås av automatisk når alledata er vist.

Diastolisk
Puls
Nyeste data

Merknad: Hvis du trykker på knappen_START_ under visning av
data i minnet, slås enheten av.

Systolisk
Diastolisk
Puls
Siste data (eldste)

Norsk 16

Hva er IHB/AFib-indikatoren?
Hvis måleren registrerer uregelmessig hjerterytme under målingen, vises
IHB/AFib-indikatoren på displayet sammen med måleverdiene.
Merknad: Vi anbefaler at du kontakter
IHB/AFib-indikatoren vises ofte.

din

lege

hvis

Hva er AFib?
Hjertet trekkes sammen på grunn av elektriske signaler som oppstår ihjertet, og som sender blod gjennom kroppen. Atrieflimmer (AFib) oppstår
når det elektriske signalet i hjertets forkammer blir forvirret, og fører til
forstyrrelser i pulsintervallene. AFib kan føre til at blod blir stående stille ihjertet, noe som kan føre til at det dannes blodpropper. Dette er en årsak
til slag og hjerteanfall.

％IHB/AFib%IHB/AFib vises som frekvensen av den registrerte IHB/AFib-verdien.
IHB/AFib kan ikke bare registrere støy, som fysisk bevegelse, men ogsåuregelmessig hjerterytme. Derfor anbefaler vi at du kontakter legen dinhvis %IHB/AFib-nivået er høyt.
Antall registrerte
IHB/AFib-forekomster i minnet
x 100 [%]
%IHB/AFib =Totalt antall
Visning av %IHB/AFib: %IHB/AFib vises når gjennomsnittsverdiene vises.
(Se “2. Hente frem data“ i “Driftsmodus“)%IHB/AFib vises ikke hvis antall verdier i minnet
er seks eller lavere.
Visning av gjennomsnittsverdierNivå 0Nivå 1Nivå 2Nivå 3%IHB/AFib=0 %IHB/AFib=1 - 9 %IHB/AFib=10 - 24 %IHB/AFib=25 - 100
Vises ikke

Norsk 17

Trykkstolpeindikator
Indikatoren overvåker fremdriften i trykket under måling.
Trykk bygges opp

Luft slippes ut

Oppblåsing pågår

Måling pågår
Oppblåsing fullført

WHO-klassifiseringsindikator
Hvert
segmenti
stolpeindikatoren
samsvarer med blodtrykksklassifiseringen
fra WHO (Verdens helseorganisasjon), som
er beskrevet på neste side.
WHO-klassifiseringsindikator
Alvorlig hypertensjon
Moderat hypertensjon
Mild hypertensjon
Høy normal
NormalOptimalt

Eksempel:

Moderat hypertensjon: Indikatoren viser et segment basert på
gjeldende data, som samsvarer med
WHO-klassifiseringen.

Mild hypertensjon

Høy normal

Om blodtrykk
Hva er blodtrykk?
Blodtrykk er kraften som blodet utøver mot veggene i arteriene. Systolisk
trykk forekommer når hjertet trekker seg sammen. Diastolisk trykk
forekommer når hjertet utvider seg. Blodtrykk måles i mm kvikksølv(mmHg). Det naturlige blodtrykket til en person representeres av det
grunnleggende trykket, som måles tidlig om morgenen mens man fortsatt
hviler, og før man spiser.
Norsk 18

Hva er hypertensjon, og hvordan kan det kontrolleres?
Hypertensjon er unormalt høyt blodtrykk i arteriene. Ubehandlet kan detforårsake mange helseproblemer, inkludert slag og hjerteanfall.
Hypertensjon kan kontrolleres ved å endre livsstil, unngå stress, samt ved
medisinering under tilsyn av en lege.
For å forebygge hypertensjon eller holde det under kontroll:
Ikke røyk
Reduser inntaket av salt og fett
Hold riktig vekt

Mosjoner regelmessig
Få utført regelmessige helsekontroller

Hvorfor måle blodtrykket hjemme?
Hvis blodtrykket måles ved en klinikk eller et legekontor, kan det føre tiluro, som gir høyere verdier, 25 til 30 mmHg høyere enn det som måles
hjemme. Måling hjemme reduserer virkningen av ekstern påvirkning på
blodtrykksverdiene, er et supplement til legens verdier, og gir en mer
nøyaktig og fullstendig blodtrykkshistorikk.

WHOs klassifisering av blodtrykk

Variasjoner i blodtrykket

Diastolisk blodtrykk

Diagrammet
viser
Referansemateriell: Journal of Hypertension 1999, Vol 17 No.2
standarder for evaluering mmHg
Hypertensjon grad 3 (alvorlig)
av høyt blodtrykk uten
110
Hypertensjon grad 2 (moderat)
hensyntilalder,
er
105
100
utarbeidet av Verdens
Hypertensjon grad 1 (mild)
95
helseorganisasjon (WHO).
90
85
80

Høy-normal
NormalBlodtrykk

Blodtrykket til en person
Optimaltvarierer sterkt på dags- og
120 130 140 150 160 170 180
Systolisk blodtrykk
mmHg
sesongbasis.
Detkanvariere med 30 til 50
mmHg på grunn av ulike tilstander i løpet av dagen. Hos personer med
høyt blodtrykk er variasjonene enda større. Blodtrykket stiger normalt ved
arbeid eller aktivitet, og faller til det laveste nivået mens man sover. Vær
derfor ikke for bekymret over resultatene i en enkeltmåling.
Foreta målinger på samme tidspunkt
Typiske variasjoner i løpet av en dag(Målt hvert femte minutt)
hver dag med fremgangsmåten som er
Systolisk
beskrevet i denne håndboken for å finneditt normale blodtrykk. Regelmessige
avlesninger
girmer
omfattende
blodtrykkshistorikk. Pass på å notere
dato og klokkeslett når du noterer
Søvn
Diastolisk
blodtrykket. Rådfør deg med legen din forå tolke blodtrykksverdiene dine.
PM
AM
Tid

Norsk 19

Feilsøking
ProblemMulig årsak
Ingenting vises
Batteriene er utladet.
på displayet
selv om
Batteripolene vender
enheten er slått
ikke riktig vei.
på.
Batterispenningen er for
lav.
(Symbolet for LAVT
Mansjetten
BATTERINIVÅ) blinker.
blåses ikke opp.
Symbolet vises ikke hvis
batteriene er helt
utladet.
Mansjetten sitter ikke
som den skal.

Anbefalt tiltak
Erstatt alle batteriene med nye
batterier.
Installer batteriene igjen med
pluss- og minuspolene som
angitt på batterirommet.

Erstatt alle batteriene med nye
batterier.

Sett mansjetten riktig på.

Pass på å sitte helt stille ogunngå å snakke undermålingen.
Enheten måler
Sitt komfortabelt og stille.
ikke blodtrykket.
Legg armen på et bord med
Verdiene er for Mansjettens posisjon erhåndflaten vendt opp og
høye eller for
ikke riktig.
mansjetten på høyde medlave.
hjertet.
Hvis du har svært svak elleruregelmessig puls, kan detvære vanskelig for enheten å
fastslå blodtrykket ditt.
Verdien avviker fra
verdien som er målt ved Se “Hvorfor måle blodtrykket
en klinikk eller ethjemme?“.
legekontor.
Annet
Ta ut batteriene. Sett inn
batteriene igjen på riktig måte,og foreta en ny måling.
Merknad: Kontakt forhandleren hvis tiltakene som er beskrevet over ikke løser
problemet. Ikke prøv å åpne eller reparere dette produktet. Alleforsøk på å gjøre det fører til at garantien blir ugyldig.
Du har bevegd påarmen under målingen.

Norsk 20

Vedlikehold
Enheten må ikke åpnes. Den inneholder skjøre elektriske komponenterog en intrikat luftenhet som kan bli skadet. Hvis problemet ikke kan
utbedres ved å følge instruksjonene for feilsøking, kontakter du en
autorisert forhandler eller vår kundeserviceavdeling. Kundeservice hos
A&D tilbyr teknisk informasjon, reservedeler og enheter til autoriserte
forhandlere.
Enheten er designet og fremstilt for å vare lenge. Det anbefales imidlertidå få enheten kontrollert annethvert år for å sikre riktig funksjon og
presisjon. Kontakt den lokale autoriserte forhandleren eller A&D for å få
utført vedlikehold.

Tekniske spesifikasjonerTypeMålemetodeMåleområde

Målepresisjon
Strømforsyning
Antall målinger

Klassifisering
Klinisk test
EMD
Minne
Driftsvilkår
Transport-/lagringsvilkår
Mål
Vekt

UA-767S
Oscillometrisk måling
Trykk:
0 - 299 mmHg
Systolisk trykk: 60 - 279 mmHg
Diastolisk trykk: 40 - 200 mmHg
Puls:
40 - 180 hjerteslag/minutt
Trykk:±3 mmHgPuls:±5%
4 x 1,5 V batterier (R6P, LR6 eller AA) eller
Vekselstrømadapter (TB-233C) (ikke inkludert)Ca. 700 ganger
LR6 (alkaliske batterier)Ca. 200 ganger
R6P (manganbatterier)
Ved en trykkverdi på 180 mmHg ved
romtemperatur på 23 °C.
Internt drevet ME-utstyr (forsyning med batterier) /klasse II (forsyning med strømadapter)
Kontinuerlig driftsmodus
Ifølge ISO81060-2：2013
I den kliniske valideringsstudien ble K5 brukt på 85
forsøkspersoner for å fastslå diastolisk blodtrykk.
IEC 60601-1-2: 2014
Siste 60 målinger+10 til +40 °C / 15 til 85 %RH / 800 til 1060 hPa-20 til +60 °C / 10 til 95 %RH / 700 til 1060 hPa
Ca. 140 [B] x 60 [H] x 105 [D] mm
Ca. 245 g uten batterier
Norsk 21

Inntrengningsbeskyttelse Enhet:

IP20

Anvendt delLevetid

Mansjett Type BF
Enhet:
5 år (ved bruk seks ganger daglig)
Mansjett: 2 år (ved bruk seks ganger daglig)
Vekselstrømadapter: 5 år (ved bruk seks ganger
daglig)Vekselstrømadapter (tilbehør)
Adapteren brukes til å koble enheten til en strømkilde i boligen. Kontakt den
lokale A&D-forhandleren for anskaffelse. Vekselstrømadapteren må
inspiseres eller skiftes periodisk.
TB-233C
Kontakt den lokale A&D-forhandleren for anskaffelse.
Vekselstrømadapteren må inspiseres eller skiftes
periodisk.
Trykte symboler på strømadapteren
SymbolerFunksjon/betydning
Kun for bruk innendørs
Enhet i klasse II
Termisk sikring
Sikring
Symbol for EC-direktivet
Symbol for EAC-sertifisering
Vekselstrømadapterpluggens polaritet
Tilbehør som selges separat
Mansjett Katalognummer
Mansjettstørrelse
Armstørrelse
CUF-F-LA
Stor mansjett for voksne
31 cm til 45 cm
CUF-I
Universalmansjett
22 cm til 42 cm
CUF-F-A
Mansjett for voksne
22 cm til 32 cm
Armstørrelse: Armens omkrets ved biceps.
Vekselstrømadapter

Katalognummer
TB-233C

Plugg (utløpstype)Type C

Merknad: Spesifikasjonene kan endres uten varsel.
IP-klassifisering er innkapslingens beskyttelsesgrad i henhold til
IEC 60529. Denne enheten er beskyttet mot faste gjenstander med
12 mm diameter og større, for eksempel fingre. Denne enheten er
ikke vanntett.

Norsk 22

Sisällys
Hyvät asiakkaat .................................................................... 2
Ennakkohuomautukset ......................................................... 2
Varotoimet ........................................................................... 2
Osien tunnistus .................................................................... 5
Symbolit ............................................................................... 6
Toimintatila .......................................................................... 8
Monitorin käyttö ................................................................... 9
Paristojen asennus/vaihto ................................................ 9
Ilmaletkun liittäminen .................................................... 10
Vaihtovirtasovittimen liittäminen.................................... 10
Oikean mansettikoon valitseminen ................................. 11
Käsivarren mansetin asettaminen ................................... 11
Tarkkojen mittausten tekeminen .................................... 13
Mittaus .......................................................................... 13
Mittauksen jälkeen ......................................................... 13
Mittaukset .......................................................................... 14
Tavallinen mittaus ......................................................... 14
Mittaus halutulla systolisella paineella............................ 15
Tarkkaa mittausta koskevia huomautuksia ..................... 15
Tietojen hakeminen muistista ............................................. 16
Mikä on IHB/AFib-indikaattori? ........................................... 17
Mikä on AFib? ..................................................................... 17％IHB/AFib .......................................................................... 17
Paineen palkki-indikaattori ................................................. 18
WHO-luokitusindikaattori ................................................... 18
Tietoja verenpaineesta ........................................................ 18
Mikä on verenpaine? ...................................................... 18
Mitä kohonnut verenpaine tarkoittaa ja
kuinka sitä hoidetaan? ................................................... 19
Miksi verenpainetta kannattaa mitata kotona? ................ 19
WHO:n verenpaineluokitus ............................................. 19
Verenpaineen vaihtelut .................................................. 19
Vianmääritys ....................................................................... 20
Huolto ................................................................................ 21
Tekniset tiedot ................................................................... 21

Suomi 1Hyvät asiakkaat
Onnittelut!
Olettehankkineet
A&D:nhuipputekniikkaa
edustavan
verenpainemonitorin. Tämä helppokäyttöinen ja tarkka laite helpottaa
päivittäistä verenpainehoitoa.
Suosittelemme lukemaan tämän oppaan huolellisesti ennen laitteen
käyttöä ensimmäistä kertaa.

Ennakkohuomautukset
Tämä laite on lääkinnällisiä laitteita koskevan neuvoston direktiivin 93/42 ETY-yhdenmukaisuusmerkintä. (0123: Numero
mukainen. Tämän osoittaa
viittaa asianomaiseen ilmoitettuun laitokseen)
Laite on tarkoitettu aikuisille, ei vastasyntyneille tai pikkulapsille tehtyihin
mittauksiin.
Käyttöympäristö.
Laiteontarkoitettu
ostajanomaan
käyttöön
kotihoitoympäristöön.
Tämä laite on tarkoitettu ihmisten verenpaineen ja sydämen lyöntitiheyden
diagnostiseen mittaamiseen.

Varotoimet
Tämän laitteen valmistuksessa on käytetty tarkkuuskomponentteja. Laitteen
altistuminen äärimmäisille lämpötiloille, kosteudelle, suoralle auringonvalolle,
iskuille tai pölylle pitää estää.
Puhdista laite ja mansetti kuivalla, pehmeällä liinalla tai vedellä kostutetulla
liinalla ja neutraalilla pesuaineella. Älä koskaan käytä alkoholia, bentseeniä,
ohennusainetta tai muita voimakkaita kemikaaleja laitteen tai mansetin
puhdistamiseen.
Vältä mansetin taittamista tiukalle tai letkun säilyttämistä pitkiä aikoja tiukkaan
käärittynä, jottei osien käyttöikä lyhene.
Varmista, etteivät vauvat tai pikkulapset voi kuristua letkuun ja johtoon.
Älä kierrä ilmaletkua mittauksen aikana. Se voi aiheuttaa loukkaantumisen
mansetin jatkuvan paineen vuoksi.
Laite ja mansetti eivät ole vedenkestäviä. Älä anna sateen, hien tai veden
kastella laitetta tai mansettia.
Mittaus voi vääristyä, jos laitetta käytetään lähellä televisiota, mikroaaltouunia,
matkapuhelinta, röntgensäteitä tai muita laitteita, joilla on voimakas
sähkökenttä.
Langattomat viestintälaitteet, kuten kotiverkkolaitteet, matkapuhelimet,
langattomat puhelimet ja niiden tukiasemat sekä radiopuhelimet voivat
Suomi 2

vaikuttaa verenpainemonitoriin. Tämän vuoksi näihin laitteisiin on pidettävä
vähintään 30 cm etäisyys.
Kun käytät laitetta uudelleen, varmista, että se on puhdas.
Käytettyjä laitteita, osia ja paristoja ei saa käsitellä tavallisena
kotitalousjätteenä. Ne on hävitettävä voimassa olevien paikallisten säädösten
mukaan.
Kun käytössä on vaihtovirtasovitin, varmista, että vaihtovirtasovittimen voi
tarvittaessa irrottaa nopeasti pistorasiasta.
Älä tee laitteeseen muutoksia. Se voi johtaa onnettomuuksiin tai laitteen
vaurioitumiseen.
Verenpaineen mittaamiseksi mansetin pitää puristaa käsivartta tarpeeksi
voimakkaasti niin, että veren virtaus valtimoon keskeytyy väliaikaisesti. Tämä
voi aiheuttaa kipua, tunnottomuutta tai väliaikaisen punaisen jäljen käsivarteen.
Näin tapahtuu etenkin silloin, jos mittaus tehdään useamman kerran peräkkäin.
Kipu, tunnottomuus ja punaiset jäljet häviävät ajan myötä.
Verenpaineen mittaaminen liian usein voi olla haitallista veren virtauksen
häiriintymisen vuoksi. Kun laitetta käytetään toistuvasti, tarkista, ettei sen
käyttö johda veren virtauksen pitkäaikaiseen heikentymiseen.
Jos sinulle on tehty mastektomia, kysy lääkäriltä ohjeita ennen laitteen käyttöä.
Älä anna lasten käyttää laitetta ilman valvontaa, äläkä käytä laitetta paikassa,
jossa se on pikkulasten ulottuvilla. Se voi johtaa onnettomuuksiin tai vaurioihin.
Laite sisältää pieniä osia. Ne voivat aiheuttaa tukehtumisvaaran, mikäli
pikkulapset vahingossa nielaisevat ne.
Kytke vaihtovirtasovitin irti, jos sitä ei käytetä mittauksen aikana.
Muiden kuin tässä oppaassa mainittujen lisävarusteiden käyttö voi olla
turvallisuusriski.
Jos paristo joutuu oikosulkuun, se voi kuumentua ja aiheuttaa palovammoja.
Anna laitteen sopeutua ympäristöön ennen sen käyttöä (noin tunnin ajan).
Kliinisiä testauksia ei ole tehty vastasyntyneiden lasten tai raskaana olevien
naisten osalta. Älä käytä laitetta vastasyntyneiden lasten tai raskaana olevien
naisten hoitoon.
Älä koske paristoihin, DC-liittimeen ja potilaaseen samanaikaisesti. Se voi
aiheuttaa sähköiskun.
Älä täytä mansettia, jos sitä ei ole kääritty olkavarren ympärille.

Kontraindikaatiot

Seuraavaksi kuvataan varotoimet laitteen oikeaa käyttöä varten.
Älä laita mansettia käsivarteen, jossa on jo toinen sähköinen lääkinnällinen
laite. Laite ei silloin ehkä toimi kunnolla.
Terveysongelmien välttämiseksi henkilöiden, joilla on käsivarren vakava
verenkiertohäiriö, tulee kysyä lääkäriltä ohjeita ennen laitteen käyttöä.
Älä tee itse diagnoosia mittaustulosten perusteella äläkä aloita hoitoa
omatoimisesti. Pyydä aina lääkäriä arvioimaan tulokset ja hoidontarve.
Älä laita mansettia käsivarteen, jossa on parantumaton haava.
Suomi 3Älä laita mansettia käsivarteen, johon tehdään suonensisäistä tiputusta taiverensiirtoa. Se voi johtaa vammoihin tai onnettomuuksiin.
Älä käytä laitetta tilassa, jossa on syttyviä kaasuja, kuten anestesiakaasuja. Se
voi aiheuttaa räjähdyksen.
Älä käytä laitetta ympäristöissä, joissa on suuria happipitoisuuksia, kuten
korkeapaineisessa happikammiossa tai happiteltassa. Se voi johtaa tulipaloon
tai räjähdykseen.

Suomi 4

Osien tunnistus
Näyttö

Vaihtovirtasovittimen
pistokeDC-liitin

Ilmaliitospistoke

START-painike
Ilmaliitäntä

Ilmaletku

Oikea sopivuusalue

Valtimon paikkamerkki
Osoitinmerkki

Käsivarren
mansetti

Paristolokero

Paristolokeron
kansi

1,5 V:n paristot(R6P, LR6 tai AA)

Näyttö

PM-tietojen
MUISTIIHB/AFib-symboli
Systolinen
paine%IHB/AFib
Mansetin sopivuusvirheen symboli

Diastolinen
paine

Liikkeestä johtuvan virheen symboli
Sydänmerkki

Sydämenlyöntitiheys

WHO-luokitusindikaattorija paineen palkki-indikaattori
Pariston indikaattori
Suomi 5

Symbolit
Laitteen laukkuun merkityt symbolit
Symbolit
SYS
DIA
PUL

Toiminta/Tarkoitus
Valmiustila ja laitteen päälle kytkentä.
Systolinen verenpaine yksikössä mmHg
Diastolinen verenpaine yksikössä mmHg
Sykettä minuutissa
Pariston asennusohje
Tasavirta
Tyyppi BF: Laitteessa, mansetissa ja letkuissa on suojaus
sähköiskuja vastaan.
EY:n direktiivin mukaisen lääkinnällisen laitteen merkintä
Edustaja EU:ssa
Valmistaja
Valmistuspäivä

IP

Kansainvälinen suojausmerkintä
WEEE-merkintä

SN

Sarjanumero
Katso ohjekirjaa/-vihkoa
DC-liittimen napaisuus
Pidä kuivana

Näytössä näkyvät symbolit
Symbolit

Toiminta/Tarkoitus
Tulee näkyviin mittauksen aikana.
Vilkkuu, kun pulssi löytyy.

Suositeltu toimi
Mittaus on käynnissä.
Liiku mahdollisimman
vähän.

IHB/AFib-symboli
Tulee näkyviin, kun havaitaan
epäsäännöllinen sydämensyke.
Se saattaa syttyä, jos havaitaan hienoista
tärinää, kuten vapinaa tai puistatusta.
Mittaustuloksen arvo voi
olla virheellinen.
Tulee näkyviin, kun havaitaan vartalon tai
Yritä mitata uudelleen.
käsivarren liike.
Pysy mittauksen aikana
liikkumatta.
Suomi 6

Symbolit
Symbolit

Toiminta/Tarkoitus

Suositeltu toimi

Tulee näkyviin mittauksen aikana, josmansetti on kiinnitetty löysästi

Mittaustuloksen arvo voi ollavirheellinen. Kiinnitä
mansetti oikein ja yritä mitata
uudelleen.

IHB/AFib-havaintojen määrä muistissa%IHB/AFib =Muistissa olevien
IHB/AFibs-havai
ntojen määrä

x100 [%]

Määrä yhteensä

Aiemmat muistiin (MEMORY)
tallennetut mittaukset.
Keskiarvotiedot
PARISTO TÄYSI
Pariston virtaindikaattori mittauksen
aikana.
PARISTO VÄHISSÄ
Tämä vilkkuu, kun paristo on vähissä.
Epävakaa verenpaine mittauksen
aikaisen liikkumisen vuoksi.

Vaihda kaikki paristot uusiin,
kun merkki vilkkuu.
Yritä mitata uudelleen.
Pysy mittauksen aikana
täysin liikkumatta.

Systolisen ja diastolisen arvon välinen
ero on enintään 10 mmHg.
Painearvo ei suurentunut täytön
aikana.
Kiinnitä mansetti oikein ja
yritä mitata uudelleen.
Mansettia ei ole asetettu oikein.
PUL DISPLAY ERROR
Pulssia ei havaita oikein.

Verenpainemonitorin sisäinen virhe

Suomi 7

Poista paristot, paina
START -painiketta ja asenna
sitten paristot uudelleen. Josvirhe näytetään yhä, otayhteys jälleenmyyjään.

Toimintatila
1. Tavallinen mittaus
Paina START -painiketta. Verenpaine mitataan ja tiedot tallennetaan muistiin.
Laite voi tallentaa muistiin 60 viimeisintä mittausta.

2. Tietojen hakeminen
Kun mitään ei näy, paina START -painiketta ja
pidä se pohjassa.

Vapauta
painike,keskiarvotiedot.

kun

näkyviin

Valmiustilassa,
paina
ja
pidä
painettuna
START

painiketulevatTietojen numero ja tallennetut tiedot näytetään
automaattisesti
järjestyksessä
viimeisestä
mittauksesta alkaen.

Tarkemmat tiedot tietojen muistista hakemisesta
löytyvät osiosta ”Tietojen hakeminen muistista”.

3. Kaikkien muistissa olevien tietojen poistaminen
Kun laite sammutetaan, paina
START -painiketta, kunnes ”
Tyhjennä tiedot valitsemalla ”” tulee näyttöön.
”.

-merkki vilkkuu.
Tiedot on tyhjennetty, kun
Laite sammuu automaattisesti.

4. Mittaus halutulla systolisella paineella
Katso sivulta 15 tiedot mittaamisesta halutulla systolisella paineella.
Suomi 8START

Monitorin käyttö
Paristojen asennus/vaihto

Vaihe 1

1. Irrota paristolokeron kansi.

Vaihe 3

2. Poista käytetyt paristot ja asenna
uudet paristot paristolokeroon kuvan
mukaisesti.
Varmista,että
napaisuudet (+ ja -) asetetaan oikein.
Käytä vain paristoja R6P, LR6 tai AA.

Vaihe 1

Vaihe 2

3. Asenna paristolokeron kansi.

HUOMAUTUS
Laita paristot paristolokeroon kuvatulla tavalla. Jos ne laitetaan väärin, laite ei
toimi.
Kun(PARISTO VÄHISSÄ -merkki) vilkkuu näytössä, vaihda kaikki paristot
uusiin. Älä sekoita vanhoja ja uusia paristoja. Se voi lyhentää paristojen
käyttöikää tai aiheuttaa laitteeseen toimintahäiriön.
Vaihda paristot, kun laitteen sammumisesta on kulunut kaksi sekuntia tai
enemmän.
(PARISTO VÄHISSÄ -merkki) näkyy, vaikka paristot on vaihdettu, teeJosverenpainemittaus. Laite voi tällöin tunnistaa uudet paristot.
(PARISTO VÄHISSÄ -merkki) ei tule näkyviin, kun paristot ovat tyhjät.
Paristojen käyttöikä vaihtelee ympäristön lämpötilan mukaan. Se voi olla lyhempi
alhaisissa lämpötiloissa. Yleensä neljä uutta LR6-paristoa kestävät noin yhden
vuoden, kun niitä käytetään mittaukseen kahdesti päivässä.
Käytä vain määritettyjä paristoja. Laitteen mukana toimitetut paristot ontarkoitettu monitorin toiminnan tarkastamiseen, joten niiden käyttöikä voi ollalyhyempi.
Poista paristot, jos laitetta ei tulla käyttämään pitkään aikaan.
Paristot saattavat vuotaa ja aiheuttaa laitteeseen toimintahäiriön.

Suomi 9

Monitorin käyttö
Ilmaliitäntärliitäntä

Ilmaletkun liittäminen
Asenna ilmaliitospistoke tiukasti
ilmaliitäntään.

Ilmaliitospistoke

Vaihtovirtasovittimen liittäminen
Kytke vaihtovirtasovittimen pistoke
DC-liittimeen.
Liitä
sittenvaihtovirtasovitin
pistorasiaan.
Käytä määritettyä vaihtovirtasovitinta.
(Katso sivu 22.)

DC-liitin
Vaihtovirtasovittimen pistoke

Kun irrotat vaihtovirtasovittimen pistorasiasta, tartu vaihtovirtasovittimen
runkoon ja vedä se ulos pistorasiasta.
Kun
irrotatvaihtovirtasovittimen
verenpainemonitorista,
vaihtovirtasovittimen pistokkeeseen ja vedä se ulos monitorista.

Suomi 10tartu

Monitorin käyttö
Oikean mansettikoon valitseminen
On tärkeää käyttää oikeankokoista mansettia, jotta mittaustulokset olisivat
tarkkoja. Jos mansetti ei ole oikeankokoinen, verenpainemittauksen tulos voi ollavirheellinen.
Käsivarren koko on merkitty jokaiseen mansettiin.
Mansettiin merkitty osoitin
ja oikea sopivuusalue kertovat, onko käytössä
oikea mansetti. (Katso seuraavalta sivulta ”Mansettiin merkityt symbolit”.)
Jos osoitin
osuu alueen ulkopuolelle, ota yhteyttä paikalliseen jälleenmyyjään
ja hanki toinen mansetti.
Käsivarren mansetti on kuluva tuote. Jos se on kulunut, hanki tilalle uusi.
Käsivarren koko
31 cm–45 cm

Suositeltu mansetin koko
Kataloginumero
Suuri aikuisten mansetti
CUF-F-LA
Laajan kokoalueen
22 cm–42 cm
CUF-I
mansetti
22 cm–32 cm
Aikuisten mansetti
CUF-F-A
Käsivarren koko:
Ympärysmitta hauiksen kohdalla.
Huom.:
Mallia UA-767S ei ole suunniteltu pienen mansetin käyttöön.

Käsivarren mansetin asettaminen
1. Kääri mansetti olkavarren ympärille,
noin 1–2 cm kyynärpään sisäpuolen
yläpuolelle, kuten kuvassa.
Laita mansetti suoraan iholle. Vaatteet
voivat heikentää pulssin tuntumista ja
aiheuttaa siksi mittausvirheen.

Älä rullaa paidanhihaa kireälle.
Mansetti
1–2 cmIlmaletku

2. Jos tiukalle kääritty hiha puristaa
olkavartta, tarkkoja mittaustuloksia eiehkä saada.

Valtimon paikkamerkki
Kiinnitäkangaskiinnike
tiukasti.

osuu oikean
3. Varmista, että osoitin
sopivuusalueen sisään.

Osoitin

Suomi 11

ja oikea sopivuusalue

Monitorin käyttö
Mansettiin merkityt symbolit
Symbolit
Toiminta/Tarkoitus

Suositeltu toimi
Aseta -merkki olkavarren
valtimonpäälletaikäsivarren
sisäpuolellasamaan linjaan nimettömän
kanssa.

Valtimon paikkamerkki
Osoitin
Kataloginumero
Aikuisten mansetin oikea sopivuusalue.
Merkitty aikuisten mansettiin.

▲
REF

Käytä
suurtaaikuistenmansettia
aikuisten
Alueen yli, merkitty aikuisten mansettiin
mansetintailaajanja laajan kokoalueen mansettiin.
kokoalueen
mansetinsijaan.
Laajan kokoalueen mansetin oikea
sopivuusalue.
Merkittylaajankokoalueen mansettiin.
Suuren aikuisten mansetin oikea
sopivuusalue.
Merkitty
suureenaikuisten mansettiin.
Alueenalle,
merkitty
aikuistenmansettiinjalaajankokoalueen
mansettiin.
Käytä aikuisten mansettia
Alueenallesuuressa
aikuistensuuren aikuisten mansetin
mansetissa.
sijaan.
Eränumero

Suuri aikuisten mansettiA

Oikea sopivuusalue

L

Laajan kokoalueen mansetti

S

W

L

A

L

Aikuisten mansettiS

Suomi 12

Monitorin käyttö
Tarkkojen mittausten tekeminen
Verenpainemittaus antaa tarkimmat tulokset, kun toimitaan seuraavasti:
Istu mukavasti tuoliin. Laita käsivarsi lepäämään pöydälle. Älä laita jalkoja ristiin.
Pidä jalat tasaisesti lattiassa ja suorista selkä.
Rentoudu viidestä kymmeneen minuuttia ennen mittauksen tekemistä.
Laita mansetin keskikohta sydämen tasolle.
Pysy paikallasi ja puhumatta mittauksen ajan.
Älä mittaa välittömästi fyysisen rasituksen tai kylvyn jälkeen. Lepää
kaksikymmentä tai kolmekymmentä minuuttia ennen mittauksen tekemistä.
Pyri mittaamaan verenpaine joka päivä samaan kellonaikaan.
Mittaus
On normaalia, että mansetti tuntuu mittauksen aikana hyvin tiukalta. (Äläsäikähdä.)

Mittauksen jälkeen
Sammuta laite mittauksen jälkeen painamalla START -painiketta.
Poista mansetti ja kirjaa tiedot muistiin.
Huom.: Laitteessa on automaattinen sammutustoiminto, joka sammuttaa laitteen
noin minuutin kuluttua mittauksesta.
Odota vähintään kolme minuuttia ennen mittauksen tekemistä samalle
henkilölle.

Suomi 13

Mittaukset
Lue ennen mittausta seuraavan sivun osio ”Tarkkaa mittausta koskevia
huomautuksia”.

Tavallinen mittaus
1.

Laita mansetti käsivarteen (mieluiten vasempaan).
Istu hiljaa mittauksen ajan.
Sydämen tasolla

2.

Paina START -painiketta.
Kaikki näyttösegmentit ovat näkyvissä.
Nolla (0) näkyy hetken vilkkuvana.
Näyttö
muuttuu
mittauksen
alkaessa
oikeanpuoleisessa
kuvassa
osoitetulla
tavalla. Mansetti alkaa täyttyä. On normaalia,
että mansetti tuntuu hyvin tiukalta. Täytön
aikana näytön vasemmassa reunassa näkyy
paineen palkki-indikaattori.

3.

Paina

START

Kaikki näytön
alueet ovat näkyvissä

Huom.: Täytön voi keskeyttää milloin
tahansa
painamalla
uudelleen
START-painiketta.

Nolla-näyttö
Täytön aloitus

Kun mansetti on täytetty, se alkaa tyhjentyä
automaattisesti ja(sydänmerkki) vilkkuu
ilmoittaen, että mittaus on käynnissä. Kun
pulssi löytyy, merkki vilkkuu jokaisella
sykkeellä.

Paineistus

Huom.: Jos riittävää painetta ei saavuteta,mansetti alkaa täyttyä uudelleen
automaattisesti.
Katso uudelleen täyttämisen estäminen
seuraavan sivun osiosta ”Mittaus halutulla
systolisella paineella”.
4.

5.

Kun mittaus on valmis, systolisen ja
Mittausdiastolisen paineen lukemat sekä sydämen
käynnissä
lyöntitiheys tulevat näyttöön.
Mansettiin jäänyt ilma poistetaan ja se
Systolinen paine
tyhjenee täysin.
Sammuta laite painamalla-painiketta.

START

Huom.: Odota vähintään kolme minuuttia ennen
mittauksen tekemistä samalle henkilölle.
Suomi 14

Diastolinen paine
WHO-luokitus
Sydämen lyöntitiheys
Poistaa jäljelle jääneen
ilman automaattisestiMittaukset
Mittaus halutulla systolisella paineella
Malli UA-767S on suunniteltu etsimään pulssi ja täyttämään mansetti
systolisen paineen tasolle automaattisesti.
Käytä tätä menetelmää, jos mansetti täyttyy toistuvasti uudelleen tai jos
tuloksia ei näytetä, vaikka paine laskee arvoon 20 mmHg tai alhaisemmaksi.
1.

Laita mansetti käsivarteen sydämen tasolle(mieluiten vasempaan).

Sydämen tasolla

2.

Paina

Paina
painiketta

3.

Kun
nolla
vilkkuu,
paina
START-painiketta ja pidä se painettuna,kunnes näkyviin tulee lukema, joka on
noin 30–40 mmHg korkeampi kuin
odotettu systolinen paine.

4.

Kun haluttu numero saavutetaan, aloita
mittaus vapauttamalla START-painike.
Jatka verenpaineen mittausta edellisellä
sivulla kuvatulla tavalla.

START -painiketta.

START

Vapauta painike
halutun systolisen
paineen kohdalla

Paina ja pidä
painettuna painike

Tarkkaa mittausta koskevia huomautuksia

START

Katso mittauksen tiedot
edelliseltä sivulta

Istu alas mukavaan asentoon. Laita käsivarsi pöydälle niin, että kämmen on
ylöspäin ja mansetti on sydämen tasolla.
Rentoudu viidestä kymmeneen minuuttia ennen mittauksen tekemistä. Jos olet
kiihtynyt tai henkisesti rasittunut, tämä näkyy mittauksessa: mitattu verenpaine
on normaalia korkeampi (tai matalampi) ja pulssi on yleensä tavallista
nopeampi.
Yksilön verenpaine vaihtelee jatkuvasti riippuen siitä, mitä hän tekee tai mitä
hän on syönyt. Juomilla voi olla voimakas ja nopea vaikutus verenpaineeseen.
Laitteen tekemät mittaukset pohjautuvat sydämen sykkeeseen. Jos syke on
erittäin heikko tai epäsäännöllinen, laitteen voi olla vaikea mitata verenpainetta.
Jos laite huomaa epänormaalit olosuhteet, se keskeyttää mittauksen ja näyttää
virhesymbolin. Katso sivulta 7 symboleiden kuvaus.
Tämä verenpainemonitori on tarkoitettu aikuisten käyttöön. Kysy ohjeita
lääkäriltä ennen laitteen käyttöä lapsen verenpaineen mittaamiseen. Lapset
eivät saa käyttää tätä laitetta ilman valvontaa.
Suomi 15

Liian korkeat lämpötilat tai suuri kosteus tai korkeus voivat vaikuttaa
automaattisen verenpainemonitorin toimintakykyyn.

Tietojen hakeminen muistista
Huom.:Tämä laite tallentaa muistiin 60 viimeisintä mittausta.
1. Kunmitään
ei
näy,
paina
_START_-painiketta ja pidä sepainettuna tallennettujen tietojen
hakemiseksi.

Paina japidä
painettuna START painike
Systolinen keskiarvo
Diastolinen keskiarvo

2. Vapauta painike, kun näkyviin tulevat
keskiarvotiedot.

Pulssin keskiarvo
Kaikkien tietojen
keskiarvo

3. Tietojen numero ja tallennetut tiedot
näytetään automaattisesti
järjestyksessä viimeisestä
mittauksesta alkaen.
Systolinen

4. Näyttö sammuu automaattisesti, kun
kaikki tiedot on näytetty.

Diastolinen
PulssiViimeisimmät tiedot

Huom.: Jos painat _START_-painiketta
tietojen haun aikana, laite sammuu.

Systolinen
Diastolinen
PulssiViimeisin tieto (vanhin)

Suomi 16

Mikä on IHB/AFib-indikaattori?
Jos monitori havaitsee epäsäännöllisen rytmin mittausten
IHB/AFib-indikaattori tulee näyttöön mittausarvojen kanssa.
Huom.: Kehotamme
ottamaanyhteyttä
lääkäriin,
IHB/AFib-indikaattori näytetään toistuvasti.

aikana,jos

Mikä on AFib?
Sydän supistuu sydämessä esiintyvien sähköisten signaalien vuoksi ja selähettää verta kaikkialle kehoon. Eteisvärinää (AFib) syntyy, kun sähköinen
signaali muuttuu eteisessä epäselväksi saaden aikaan häiriöitä pulssirytmissä.
AFib voi muodostaa sydämeen verihyytymiä. Nämä voivat helposti synnyttää
verisuonitukoksia, jotka puolestaan voivat aiheuttaa aivohalvauksen ja
sydänkohtauksen.

％IHB/AFib%IIHB/AFib on havaittujen IHB/AFib-havaintojen taajuus.
%IHB/AFib voi havaita kohinan, esim. fyysisten liikkeiden lisäksi myös
epäsäännöllisen sydämen sykkeen. Siksi suosittelemme ottamaan yhteyttä
lääkäriin, jos %IHB/AFib-taso on korkea.
Muistissa olevien
IHB/AFibs-havaintojen määrä x 100 [%]%IHB/AFib =Määrä yhteensä
Arvon %IHB/AFib näyttö: %IHB/AFib näytetään, kun näytetään keskiarvoja.
(Katso 2. Tietojen hakeminen” kohdassa ”Toimintatila”)%IHB/AFib-lukemaa ei näytetä, jos määrä muistissa on
kuusi tai vähemmän.
Keskiarvon näyttö

Taso 0%IHB/AFib=0

Taso 1%IHB/AFib=1 - 9

Taso 2%IHB/AFib=10 - 24

Ei näytetä

Suomi 17

Taso 3%IHB/AFib=25 - 100

Paineen palkki-indikaattori
Indikaattori seuraa paineen muuttumista mittauksen aikana.
Paineistus

Ilmaa vapautetaan

Täyttö käynnissä

Mittaus käynnissä
Mansetti täynnä

WHO-luokitusindikaattori
Jokainen palkki-indikaattorin alue vastaa
WHO:n verenpaineluokitusta, joka on kuvattu
seuraavalla sivulla.
WHO-luokitusindikaattori
Erittäin korkea verenpaine
Melko korkea verenpaine
Lievästi kohonnut verenpaine
Korkea-normaali
Normaali
Ihanteellinen: Indikaattori näyttää, nykyisiin tietoihin
perustuen, alueen, joka vastaa
WHO-luokitusta.

Esimerkki:

Melko korkea verenpaine

Lievästi kohonnut verenpaine

Korkea-normaali

Tietoja verenpaineesta
Mikä on verenpaine?
Verenpaine on voima, jolla veri painaa valtimoiden seinämää. Systolinen paine
syntyy, kun sydän supistuu. Diastolinen paine syntyy, kun sydän laajenee.
Verenpaine mitataan elohopeamillimetreinä (mmHg). Yksilön luonnollista
verenpainetta edustaa peruspaine, joka mitataan heti aamulla, kun henkilö on
vielä levossa, ennen aamiaista.

Suomi 18

Mitä kohonnut verenpaine tarkoittaa ja kuinka sitä hoidetaan?
Kohonnut verenpaine tarkoittaa epätavallisen korkeaa valtimoverenpainetta,
joka hoitamattomana voi aiheuttaa monia terveysongelmia, mukaan lukien
aivohalvauksen tai sydänkohtauksen. Kohonnutta verenpainetta voi hoitaa
elämäntapamuutoksella, stressiä välttämällä sekä lääkkeillä lääkärin
valvonnassa.
Kohonneen verenpaineen ehkäiseminen tai hoitaminen:
Älä tupakoi
Harrasta säännöllistä liikuntaa
Vähennä suolaa ja rasvaa
Tarkistafyysinenkunto
säännöllisesti
Ylläpidä hyvää painoindeksiä

Miksi verenpainetta kannattaa mitata kotona?
Verenpaineen mittaaminen sairaalassa tai lääkärin vastaanotolla voi aiheuttaa
hermostuneisuutta ja johtaa lukemiin, jotka ovat 25–30 mmHg korkeammat
kuin kotona mitatut. Kotona tehdyt mittaukset minimoivat verenpainelukemiin
vaikuttavat ulkoiset tekijät, täydentävät lääkäreiden tekemiä mittauksia ja
antavat tarkemman ja kattavamman kuvan verenpaineen muutoksista ajan
kuluessa.

WHO:n verenpaineluokitus

Verenpaineen vaihtelut

Viitemateriaali: Journal of Hypertesion 1999, Vol 17 No. 2

mmHg
Diastolinen verenpaine

Maailmanterveysjärjestö(WHO)onmäärittänyt
standardit
korkeanverenpaineen
arvioinnille,
ikään katsomatta, kaaviossa
kuvatun mukaisesti.

110
105
100
95
90
85
80

Luokan 3 kohonnut verenpaine (vakava)
Luokan 2 kohonnut verenpaine (kohtalainen)
Luokan 1 kohonnut
verenpaine (lievä)
Korkea-normaali
Normaali

Verenpaine

Yksilön verenpaine vaihtelee
Ihanteellinen
huomattavasti päivittäin ja
120 130 140 150 160 170 180
Systolinen verenpaineverenpaineyksikössä mmHg
kausittain. Paine voi vaihdella
30–50 mmHg
päiväneritilanteissa. Henkilöillä, joilla on kohonnut verenpaine, vaihtelut voivat olla jopa
suurempia. Normaalisti verenpaine kohoaa työnteon tai leikkimisen aikana, ja
se laskee alimmilleen henkilön nukkuessa. Tämän vuoksi yhdestä
mittaustuloksesta ei kannata huolestua liikaa.
Tyypillinen vaihtelu päivän aikana
Voitselvittää
normaalinverenpaineesi(mitataan viiden minuutin välein)tekemällä mittauksen joka päivä samaan
kellon aikaan tässä oppaassa kuvattuja
Systoline
menetelmiä noudattaen. Säännöllisesti tehdyt
mittaukset antavat kattavamman kuvan
verenpaineen muutoksista ajan kuluessa.
Muista merkitä ylös verenpainemittauksen
päivämäärä ja kellonaika. Pyydä lääkäriä
Uni
Diastolinen
tulkitsemaan verenpainetietoja.
PM

Suomi 19

AM

KellonaikaVianmääritys
OngelmaNäytössä ei näy
mitään, vaikka
laite on päällä.

Mahdollinen syy
Paristot ovat tyhjät.
Pariston navat eivät ole
oikeilla paikoilla.

Pariston jännite on liian
alhainen.
(PARISTO VÄHISSÄ
Mansetti ei täyty. -merkki) vilkkuu. Jos
paristot ovat tyhjentyneet
täysin, merkki ei tulenäkyviin.
Mansettia ei ole asetettu
oikein.
Käsivartta tai vartaloa on
liikutettu mittauksen
aikana.
Laite ei mittaa.
Lukemat ovat
liian korkeita tai
Mansetti on väärässä
liian alhaisia.
kohdassa.

Suositeltu toimi
Vaihda kaikki paristot uusiin.
Asenna paristot uudelleen niin,
että negatiiviset ja positiiviset
navat on asetettu paristolokerossa
ilmoitetun mukaisesti.

Vaihda kaikki paristot uusiin.

Laita mansetti oikein.
Varmista, että potilas pysyy
paikallaan ja puhumatta
mittauksen ajan.
Istu mukavasti ja liikkumatta.
Laita käsivarsi pöydälle niin, ettäkämmen on ylöspäin ja mansetti
on sydämen tasolla.
Jos syke on erittäin heikko tai
epäsäännöllinen, laitteen voi ollavaikea mitata verenpainetta.

Arvo poikkeaa sairaalassa
Katso kohta ”Miksi verenpainetta
tai lääkärin vastaanotolla
kannattaa mitata kotona?”.
mitatusta arvosta.
Muu
Poista paristot. Laita ne oikein
takaisin ja yritä mitata uudelleen.
Huom.: Jos yllä kuvatut toiminnot eivät ratkaise ongelmaa, ota yhteyttä
jälleenmyyjään. Älä yritä avata tai korjata tuotetta; se voisi mitätöidä takuun.

Suomi 20

HuoltoÄlä avaa laitetta. Laitteessa on hienoja sähkökomponentteja ja monimutkainen
ilmayksikkö, jotka voisivat vahingoittua. Jos ongelmaa ei voida ratkaista
vianmääritysohjeita noudattamalla, ota yhteyttä alueesi valtuutettuun
jälleenmyyjään tai asiakaspalveluosastoomme. A&D:n asiakaspalvelu toimittaa
valtuutetuille jälleenmyyjille teknisiä tietoja, varaosia ja laitteita.
Laite on suunniteltu ja valmistettu niin, että sen käyttöikä on pitkä. Laite on
kuitenkin hyvä tarkistaa 2 vuoden välein sen asianmukaisen toiminnan ja
tarkkuuden varmistamiseksi. Ota yhteyttä alueesi valtuutettuun jälleenmyyjään
tai A&D:hen huollon tekemiseksi.

Tekniset tiedot
Tyyppi
Mittausmenetelmä
Mittausalue

UA-767S
Oskillometrinen mittaus
Paine:
0–299 mmHg
Systolinen paine: 60–279 mmHg
Diastolinen paine: 40–200 mmHg
Pulssi:40–180 sykettä/minuutti
Mittaustarkkuus
Paine:±3 mmHg
Pulssi:±5 %Virransyöttö
4 x 1,5 V paristot (R6P, LR6 tai AA) tai
Vaihtovirtasovitin (TB-233C) (ei sisälly toimitukseen)
Mittausten määrä
Noin 700 kertaa
LR6 (alkaliparistot)
Noin 200 kertaa
R6P (mangaaniparistot)
Painearvolla 180 mmHg huonelämpötilassa 23 °C.
Luokitus
Sähköinen lääkinnällinen laite sisäisellä virtalähteellä(paristot) / Luokka II (sovitin)
Jatkuva toimintatila
Kliininen testi
Noudattaen standardia ISO81060-2：2013
Kliinisessä validointitutkimuksessa K5:ttä käytettiin 85
koehenkilön diastolisen verenpaineen mittaamiseen.
EMD
IEC 60601-1-2: 2014
Muisti
60 viimeistä mittausta
Toimintaolosuhteet+10…+40 °C/suht. kosteus 15–85 %/800–1 060 hPa
Kuljetus-/säilytysolosuhteet -20…+60 °C /suht. kosteus 10–95 % / 700–1 060 hPa
Mitat
Noin 140 [L] x 60 [K] x 105 [S] mm
Paino
Noin 245 g, ilman paristoja
Kotelointiluokka
Laite:
IP20
Liitäntäosa
Mansetti
Tyyppi BF

Suomi 21

Käyttöikä

Laite:

5 vuotta (kun käytetään kuusi kertaa
päivässä)
Mansetti: 2 vuotta (kun käytetään kuusi kertaa
päivässä)
Vaihtovirtasovitin: 5 vuotta (kun käytetään kuusi kertaa
päivässä)
Lisävaruste, vaihtovirtasovitin
Sovittimella laite kytketään kodin virtalähteeseen. Sovittimen voi hankkia ottamalla
yhteyttä paikalliseen A&D-jälleenmyyjään. Vaihtovirtasovitin pitää tarkistaa tai
vaihtaa säännöllisesti.
TB-233C
Sovittimen voi hankkia ottamalla yhteyttä paikalliseen
A&D-jälleenmyyjään.
Vaihtovirtasovitin pitää tarkistaa tai vaihtaa
säännöllisesti.
Vaihtovirtasovittimeen merkityt symbolit
Symbolit
Toiminta/Tarkoitus
Vain sisäkäyttöön
Luokan II laite
Lämpösulake
SulakeEU-direktiivin mukaisen laitteen merkintä
EAC-sertifioidun laitteen merkintä
Vaihtovirtasovittimen
pistokkeen
napaisuus

Lisävarusteet myydään erikseen
Mansetti
Kataloginumero
Mansetin koko
Käsivarren koko
CUF-F-LA
Suuri aikuisten mansetti
31 cm–45 cm
Laajan kokoalueen
CUF-I
22 cm–42 cm
mansetti
CUF-F-A
Aikuisten mansetti
22 cm–32 cm
Käsivarren koko: Ympärysmitta hauiksen kohdalla.
Vaihtovirtasovitin

Kataloginumero
TB-233C

Pistoke(pistorasian tyyppi)
Tyyppi C

Huom.: Teknisiä tietoja voidaan muuttaa ilman ennakkoilmoitusta.
IP-luokitus on koteloiden antama suojausluokka standardin IEC 60529
mukaisesti. Laite on suojattu sellaisia vieraita esineitä vastaan, joiden
halkaisija on 12 mm tai suurempi, esim. sormet. Laitetta ei ole suojattu
vedeltä.

Suomi 22İçindekilerDeğerli Müşterilerimiz .......................................................... 2Başlangıç Açıklamaları .......................................................... 2Önlemler .............................................................................. 2Parçaların Tanımı .................................................................. 5
Semboller ............................................................................. 6Çalışma Modu ....................................................................... 8
Monitörü Kullanma ............................................................... 9
Pilleri Takma / Değiştirme ............................................... 9
Hava Hortumunu Bağlama .............................................. 10
AC Adaptörünü Bağlama ................................................ 10Doğru Manşon Boyutunu Seçme ..................................... 11Manşonu Takma ............................................................ 11Doğru Ölçümleri Yapma ................................................. 13Ölçüm ............................................................................ 13Ölçümden Sonra ............................................................ 13Ölçümler............................................................................. 14
Normal Ölçüm ............................................................... 14İstenen Sistolik Basınçla Ölçüm ...................................... 15Doğru Ölçümle İlgili Notlar............................................. 15
Bellek Verilerini Geri Çağırma.............................................. 16
IHB/AFib Göstergesi Nedir? ................................................. 17
AFib Nedir? ......................................................................... 17％IHB/AFib .......................................................................... 17
Basınç Çubuğu Göstergesi .................................................. 18
WHO Sınıflandırma Göstergesi ............................................ 18
Tansiyon Hakkında ............................................................. 18
Tansiyon Nedir? ............................................................. 18
Hipertansiyon Nedir ve Nasıl Kontrol Edilir? .................... 19
Tansiyon Neden Evde Ölçülmelidir? ................................ 19
WHO Tansiyon Sınıflandırması ........................................ 19
Tansiyon Değişimleri ..................................................... 19
Sorun giderme .................................................................... 20
Bakım ................................................................................. 21
Teknik Veriler ..................................................................... 21
Türkçe 1Değerli Müşterilerimiz
Günümüzdeki en gelişmiş monitörlerden biri olan modern A&D tansiyon
monitörünü satın aldığınız için sizi tebrik ederiz. Kullanım kolaylığı vedoğruluk için tasarlanmış bu aygıt, günlük tansiyon yönetiminizi
kolaylaştıracaktır.
Aygıtı ilk kullanımınız öncesinde bu kılavuzu dikkatlice okumanızıöneririz.

Başlangıç Açıklamaları
Bu aygıt, Tıbbi Ürünler için Avrupa Direktifi 93/42 EEC’ye uygundur. Bu,
uyumluluk işareti ile belirtilmiştir. (0123: İlgili yetkili kuruluşa
referans numarası)
Aygıt, yenidoğanlarda veya bebeklerde değil, yetişkinlerde kullanım içintasarlanmıştır.
Kullanım ortamı. Bu aygıtı evde bakım sağlanan bir ortamda kendibaşınıza kullanabilirsiniz.
Bu aygıt, tanı konulacak kişilerin tansiyon ve nabız hızını ölçmek içintasarlanmıştır.

Önlemler
Bu aygıtın üretiminde hassas parçalar kullanılmıştır. Sıcaklık uç değerleri,nem, doğrudan güneş ışığı, darbe ve tozdan kaçınılmalıdır.
Aygıtı ve manşonu kuru, yumuşak bir bezle veya su ve yumuşak
deterjanla nemlendirilmiş bir bezle temizleyin. Aygıtı veya manşonu
temizlemek için kesinlikle alkol, benzen, tiner veya başka sert kimyasallarkullanmayın.
Manşonu sıkıca katlamaktan veya hortumu uzun süre sıkı bir şekilde
bükülü tutmaktan kaçının çünkü bu tür davranışlar parçaların ömrünü
kısaltabilir.
Hortumun ve kablonun kazayla bebekleri veya küçük çocuklarıboğmamasına dikkat edin.
Ölçüm sırasında hava hortumunu bükmeyin. Bu durum, sürekli manşon
basıncı nedeniyle yaralanmaya neden olabilir.
Aygıt ve manşon suya dayanıklı değildir. Yağmur, ter ve suyun aygıtı vemanşonu kirletmesine izin vermeyin.
Aygıt televizyonlara, mikrodalga fırınlara, cep telefonlarına, X-ışını veya
güçlü elektrik alanına sahip diğer aygıtlara yakın kullanılırsa ölçümler
bozulabilir.
Türkçe2

Ev ağ aygıtları, cep telefonları, kablosuz telefonlar ve baz istasyonları,
telsizler gibi kablosuz iletişim aygıtları bu tansiyon monitörünü etkileyebilir.
Bu yüzden, bu tür aygıtlarla arada minimum 30 cm’lik bir mesafe
korunmalıdır.
Aygıt yeniden kullanılırken aygıtın temiz olduğunu doğrulayın.
Kullanılmış ekipman, parçalar ve piller normal ev atıkları olarakdeğerlendirilmez ve uygun yerel düzenlemelere göre atılmalıdır.
AC adaptörü kullanılırken, AC adaptörünün gerektiğinde hemen elektrik
prizinden çıkarılabileceğinden emin olun.
Aygıtta değişiklik yapmayın. Kazalara veya aygıtın hasar görmesine
neden olabilir.
Tansiyonu ölçmek için kol, manşon tarafından arterden kan akışını geçici
olarak durdurmaya yetecek kadar sert sıkılmalıdır. Bu durum kolda acıya,uyuşmaya veya geçici kırmızı izlere neden olabilir. Bu durum özellikleölçüm art arda tekrarlandığında görünür. Acı, uyuşma veya kırmızı izler
zamanla kaybolur.
Tansiyonu çok sık ölçmek kan akışını etkilediğinden zarara neden olabilir.
Aygıtı tekrar tekrar kullanırken aygıtın çalışmasının kan dolaşımını uzunsüre bozmadığını kontrol edin.
Mastektomi olduysanız aygıtı kullanmadan önce lütfen doktora danışın.
Çocukların aygıtı kendi başlarına kullanmasına izin vermeyin ve aygıtı
küçük çocukların erişebileceği yerlerde kullanmayın. Kazalara veya
hasara neden olabilir.
Küçük çocukların yanlışlıkla yutması durumunda boğulma tehlikesine
neden olabilen küçük parçalar içerir.
Ölçüm sırasında kullanılmadığında AC adaptörünü prizden çekin.
Bu kılavuzda belirtilmeyen aksesuarların kullanımı güvenliği tehlikeye
sokabilir.
Pil kısa devre yaparsa ısınabilir ve yanıklara neden olabilir.
Kullanmadan önce aygıtın çevre ortamına adapte olmasına izin verin(yaklaşık bir saat).
Yenidoğan bebeklerde ve hamile kadınlarda klinik test yapılmamıştır.
Yenidoğan bebeklerde veya hamile kadınlarda kullanmayın.
Pillere, DC jakına ve hastaya aynı anda dokunmayın. Bu, elektrikçarpmasıyla sonuçlanabilir.
Manşonu üst kol çevresine sarmadan şişirmeyin.

KontrendikasyonlarAşağıda aygıtın uygun kullanımı ile ilgili önlemler bulunmaktadır.
Manşonu, başka bir elektrikli tıbbi ekipmanın takılı olduğu kola takmayın.
Ekipman düzgün çalışmayabilir.
Kolunda ciddi dolaşım bozukluğu olan hastalar, tıbbi sorunlarlakarşılaşmamak için aygıtı kullanmadan önce bir doktora başvurmalıdır.
Türkçe3Ölçüm sonuçları üzerinden kendi kendinize teşhis koymayın ve tedaviyebaşlamayın. İşlemin sonuçlarının değerlendirilmesi ve tedavi için her
zaman bir doktora başvurun.
İyileşmemiş yarası olan bir kola manşonu takmayın.
İntravenöz damlama veya kan nakli yapılan kola manşonu takmayın.
Yaralanmaya veya kazaya neden olabilir.
Aygıtı anestezik gazlar gibi yanıcı gazların bulunduğu yerlerde
kullanmayın. Patlamaya neden olabilir.
Aygıtı, yüksek basınçlı oksijen odası veya oksijen çadırı gibi fazlakonsantre oksijenli ortamlarda kullanmayın. Yangına veya patlamaya
neden olabilir.

Türkçe4

Parçaların Tanımı
Ekran

Hava Bağlantı Aparatı

AC adaptörü FişiDC Jakı
START Düğmesi

Hava Soketi

Hava Hortumu

Uygun Oturtma Aralığı

Arter Konum İşaretiİşaretManşonPil Bölmesi
Pil Kapağı

1,5V Piller(R6P, LR6 veya AA)

Ekran

BELLEK

Ortalama

IHB/AFib sembolü
Sistolik
Basınç

%IHB/AFibManşon Oturma Hatası Sembolü

Diyastolik
Basınç

Hareket Hatası Sembolü
Kalp İşareti
WHO Sınıflandırma Göstergesi
ve Basınç Çubuğu Göstergesi

Nabız Hızı

Pil Göstergesi
Türkçe5

Semboller
Aygıt kasası üzerine basılmış semboller
Semboller
SYS
DIA
PUL

IP

Fonksiyon / Anlamı
Aygıtı bekleme durumuna alma ve aygıtın gücünü açma.
mmHg cinsinden sistolik tansiyon
mmHg cinsinden diyastolik tansiyon
Dakikadaki nabız
Pil takma kılavuzuDoğru akım
Tip BF: Aygıt, manşon ve boru, elektrik çarpmalarına karşı özelkoruma sağlayacak şekilde tasarlanmıştır.
EC yönergesi tıbbi aygıt etiketi
AB temsilcisiÜreticiÜretim tarihi
Uluslararası koruma sembolü
WEEE etiketi

SN

Seri numarası
Talimat kılavuzuna / kitapçığa bakın
DC jakı polaritesi
Kuru tutun

Ekranda görünen semboller
Semboller

Fonksiyon / AnlamıÖlçüm devam ederken görünür.
Nabız algılandığında yanıp söner.

Önerilen EylemÖlçüm sürüyor. Mümkün
olduğunca hareketsiz
kalın.

IHB/AFib sembolü
Düzensiz kalp atışı algılandığında
görünür.
Ürperme veya titreme gibi çok hafif
titreşim algılandığında yanabilir.
Bir vücut veya kol hareketi
algılandığında görünür.

Türkçe6

Yanlış değer
gösterilebilir.
Yeniden ölçmeyi
deneyin. Ölçüm sırasında
hareketsiz kalın.

Semboller
Semboller

Fonksiyon / AnlamıÖnerilen Eylem
Yanlış değer gösterilebilir.
Manşon gevşek takıldığında ölçüm
Manşonu doğru takın ve
yapılırken görünür
yeniden ölçmeyi deneyin.
Bellekteki IHB/AFib algılanan
algılama oranı%IHB/AFib =Bellekteki algılanan
IHB/AFib’lerin sayısı

x100 [%]

Toplam sayı

BELLEKTE depolanan öncekiölçümler.
Ortalama veri
DOLU PİL
Ölçüm sırasında pil gücü
göstergesi.

İşaret yanıp söndüğünde
tüm pilleri yenileriyledeğiştirin.
Yeniden ölçmeyi deneyin.
Ölçüm sırasında hareket nedeniyleÖlçüm sırasında tamamen
stabil olmayan tansiyon.
hareketsiz kalın.
Sistolik ve diyastolik değerler
arasında en fazla 10 mmHg farkvar.
Şişirme sırasında basınç değeri
Manşonu doğru takın ve
artmadı.
yeniden ölçmeyi deneyin.
Manşon düzgün takılmamıştır.
DÜŞÜK PİLYanıp söndüğünde pil azalmıştır.

NABIZ GÖRÜNTÜLEME HATASI
Nabız düzgün algılanmamıştır.
Tansiyon monitörü dahili hatası

Türkçe7

Pilleri çıkarıp STARTdüğmesine basın ve sonra
pilleri yeniden takın. Hata
gözükmeye devam
ediyorsa bayiye başvurun.

Çalışma Modu
1. Normal Ölçüm
START düğmesine basın. Tansiyon ölçülür ve veri bellekte depolanır. Bu

aygıt, son 60 ölçümü bellekte depolayabilir.

2. Verileri Geri Çağırma
Ekranda hiçbir şey görüntülenmediğinde
START düğmesini basılı tutun.

Bekleme modunda,şunu basılı tutun:
STARTdüğmesi

Ortalama veri görüntülenirken düğmeyi bırakın.

Veri numarası ve depolanan veriler sonölçümden itibaren otomatik olarak görüntülenir.

Verileri geri çağırma hakkında ayrıntılar için
“Bellek Verilerini Geri Çağırma bölümüne
başvurun.

3. Bellekte Depolanan Tüm Verileri Silme
Aygıtı kapatırken,““ görüntülenene kadar START düğmesini
basılı tutun.
“ öğesini seçin.
Verileri temizlemek için “işareti yanıp söndüğünde veriler temizlenir.
Aygıt otomatik olarak kapanır.

4. İstenen Sistolik Basınçla Ölçüm
İstenen sistolik basınçla ölçüm için sayfa 15’e bakın.
Türkçe8STARTMonitörü Kullanma
Pilleri Takma / Değiştirme
1. Pil kapağını çıkarın.
2. Kullanılmış pilleri çıkarın ve yeni
pilleri polaritelerin (+ ve -) doğru
olmasına dikkat ederek gösterildiği
gibi pil bölmesine takın.
Yalnızca R6P, LR6 veya AA pilleri
kullanın.
3. Pil kapağını takın.

Adım 1
Adım 3

Adım 1
Adım 2DİKKAT
Pilleri gösterildiği gibi pil bölmesine takın. Yanlış takılırsa aygıt çalışmaz.
Ekranda(DÜŞÜK PİL işareti) yanıp söndüğünde tüm pilleri yenileriyledeğiştirin. Eski ve yeni pilleri karıştırmayın. Pil ömrünü azaltabilir veya
aygıtın arızalanmasına neden olabilir.
Pilleri değiştirmek için aygıt kapatıldıktan sonra en az iki saniye bekleyin.
Piller değiştirildikten sonra(DÜŞÜK PİL işareti) görünmeye devam
ediyorsa bir tansiyon ölçümü yapın. Böylece aygıt yeni pilleri tanıyabilir.
Piller bittiğinde
(DÜŞÜK PİL işareti) görünmez.
Pil ömrü ortam sıcaklığına göre değişebilir ve düşük sıcaklıklarda kısalabilir.
Genel olarak, her gün iki kez ölçüm yapıldığında dört adet yeni LR6 pilyaklaşık bir yıl dayanır.
Yalnızca belirtilen pilleri kullanın. Aygıtla verilen piller monitör
performansını test etmek içindir ve sınırlı ömrü vardır.
Aygıt uzun süre kullanılmayacaksa pilleri çıkarın.
Piller sızıntı yapabilir ve arızaya neden olabilir.

Türkçe9

Monitörü Kullanma
Hava soketi
r soketi

Hava Hortumunu Bağlama
Hava bağlantı aparatını hava
soketine sıkıca takın.

Hava bağlantı aparatı

AC Adaptörünü Bağlama
AC adaptörü fişini DC jakına takın.
Ardından, AC adaptörünü elektrik
prizine bağlayın.
Belirtilen AC adaptörünü kullanın.
(Bkz. sayfa 22.)

DC jakı
AC adaptörü fişiAC adaptörünü elektrik prizinden çıkarırken AC adaptörü gövdesinden
tutup çekerek prizden çıkarın.
AC adaptörü fişini tansiyon monitöründen çıkarırken AC adaptörü fişini
tutup monitörden çıkarın.

Türkçe10

Monitörü KullanmaDoğru Manşon Boyutunu SeçmeDoğru manşon boyutunu kullanmak, okunan değerin doğruluğu içinönemlidir. Manşon düzgün boyutta değilse okunan değer yanlış tansiyondeğerini gösterebilir.
Kol boyutu her manşonun üzerine basılmıştır.
Manşon üzerindekiişareti ve uygun oturtma aralığı doğru manşonu
takıp takmadığınızı size belirtir. (Sonraki sayfada yer alan “Manşon üzerine
basılmış semboller” konusuna bakın.)işareti aralığın dışını gösteriyorsa yeni bir manşon satın almak için size
yakın bayiye başvurun.
Manşon, bir sarf malzemesidir. Yıpranırsa yenisini satın alın.
Kol BoyutuÖnerilen Manşon Boyutu
Katalog Numarası
31 cm - 45 cm
Büyük Yetişkin Manşonu
CUF-F-LA
22 cm - 42 cm
Geniş Aralıklı Manşon
CUF-I
22 cm - 32 cm
Yetişkin Manşonu
CUF-F-A
Kol boyutu: Pazı çevresi.
Not:
UA-767S
modeli,küçükmanşonla
kullanılmak
için
tasarlanmamıştır.

Manşonu Takma
1. Manşonu üst kol çevresine, dirsek
içinin yaklaşık 1 - 2 cm üstünden
gösterildiği gibi sarın.
Kıyafetle ölçüm, belirsiz nabza ve
dolayısıyla ölçümün hatayla
sonuçlanmasına neden
olabileceğinden manşonu doğrudan
cildin üzerine yerleştirin.

Giysinizin kolunu çok sıkı kıvırmayın.

Manşon1-2 cmHava hortumu
Arter konum işareti

2. Giysi
kolunusıkıcayukarı
kıvırmadan
kaynaklananüstkoldaki daralma okunan değerlerindoğruluğunu önleyebilir.
3.

işaretinin uygun oturtma aralığı
içini gösterdiğini onaylayın.

Türkçe11Kumaş kopçayısağlamca sabitleyin.

işaretive uygun oturtma aralığı

Monitörü KullanmaManşon üzerine basılmış semboller
Semboller Fonksiyon / AnlamıÖnerilen Eylemişaretini üst kolun
arterinde veya kolun içtarafında yüzük parmağıyla
hizalı olarak ayarlayın.

Arter konum işareti▲
REFİşaretKatalog numarasıYetişkin manşonu için uygun
oturtma aralığı.
Yetişkin manşonu üzerine basılıdır.
Yetişkin manşonu ve geniş aralıklı Yetişkin manşonu veya
genişaralıklımanşonmanşon üzerine basılı aralık yerine
büyükyetişkinüstünde.
manşonu kullanın.
Geniş aralıklı manşon için uygun
oturtma aralığı. Geniş aralıklımanşon üzerine basılıdır.
Büyük yetişkin manşonu için uygun
oturtma aralığı. Büyük yetişkin
manşonu üzerine basılıdır.
Yetişkin manşonu ve geniş aralıklımanşon üzerine basılı aralık altında.
Büyük yetişkin manşonu
Büyük yetişkin manşonu üzerinde
yerine yetişkin manşonunu
aralık altında.
kullanın.
Parti numarası

Büyük yetişkin manşonu

A

LGeniş aralıklı manşon

S

W

L

A

L

Yetişkin manşonu

S

Türkçe12

Monitörü KullanmaDoğru Ölçümleri Yapma
En doğru tansiyon ölçümü için:
Bir sandalye üzerine rahat bir şekilde oturun. Kolunuzu masanın üzerine
koyun. Bacak bacak üzerine atmayın. Ayaklarınızı zeminde düz tutun ve
sırtınızı düzleştirin.
Ölçümden önce beş ila on dakika kadar dinlenin.
Manşonun ortasını kalbinizle aynı düzeye yerleştirin.
Ölçüm sırasında hareketsiz durun ve sessiz kalın.
Fiziki egzersizden veya banyo yaptıktan hemen sonra ölçüm yapmayın.
Ölçüme başlamadan önce yirmi veya otuz dakika dinlenin.
Tansiyonunuzu her gün aynı saatte ölçmeye çalışın.

ÖlçümÖlçüm sırasında, manşonu çok sıkı olarak hissetmeniz normaldir.
(Paniklemeyin.)Ölçümden SonraÖlçümden sonra, aygıtı kapatmak için START düğmesine basın.
Manşonu çıkarın ve verilerinizi kaydedin.
Not: Aygıtta, aygıtı ölçümden yaklaşık bir dakika sonra kapatan otomatik güç
kapatma fonksiyonu vardır.
Aynı kişinin tansiyonunu ölçerken ölçümler arasında en az üç dakika
bekleyin.

Türkçe13ÖlçümlerÖlçümden önce bir sonraki sayfada yer alan “Doğru Ölçümle İlgili
Notlar“ bölümüne bakın.

Normal Ölçüm
1.

Manşonu kol üzerine yerleştirin (tercihen sol kol).
Ölçüm sırasında sessizce oturun.

2.

düğmesine basın.
Tüm ekran segmentleri görüntülenir.
Kısa bir süreliğine yanıp sönen sıfır (0)görüntülenir.
Ölçüm başladığında sağdaki şekilde
gösterildiği gibi ekran değişir. Manşon
şişmeye başlar. Manşonun çok sıktığını
hissetmeniz normaldir. Şişme sırasında
ekranın sol kenarında bir basınç çubuğu
göstergesi görüntülenir.
Not: Şişirmeyi herhangi bir zaman
durdurmak isterseniz STARTdüğmesine yeniden basın.

Kalp seviyesinde

STARTŞuna basın:

START

Tüm ekransegmentleri
görüntülenir

Sıfır gösterilirŞişirmeye başlar

3.

Şişme tamamlandığında sönme otomatik
Basınç veriliyor(kalp işareti) yanıpolarak başlar vesönerek ölçümün sürdüğünü gösterir.
Nabız algılandığında her nabız atışında
işaret yanıp söner.
Not: Uygun bir basınç sağlanmazsa aygıt
otomatik olarak yeniden şişmeye başlar.
Yeniden şişmeyi önlemek için bir sonraki
sayfada yer alan “İstenen Sistolik
Basınçla Ölçüm“ bölümüne başvurun.

4.

Ölçüm tamamlandığında sistolik vesürüyordiyastolik basınç okunan değerleri ve nabız
hızı görüntülenir.
Manşon, kalan havayı boşaltır ve tamamen
Sistolik basınçsöner.

5.

Aygıtı kapatmak içinbasın.

Ölçüm

START

Diyastolik basınçdüğmesine WHO sınıflandırması
Nabız hızı

Not: Aynı kişide ölçüm yaparken ölçümler
arasında en az üç dakika bekleyin.
Türkçe14

Kalan havayı otomatikolarak boşaltırÖlçümlerİstenen Sistolik Basınçla Ölçüm
UA-767S nabzı algılamak ve manşonu sistolik basınç düzeyine otomatikolarak şişirmek üzere tasarlanmıştır.
Bu yöntemi, tekrar tekrar yeniden şişme gerçekleştiğinde veya basıncın
20 mmHg veya daha az bir seviyeye düşmesine rağmen sonuçlar
görüntülenmediğinde kullanın.
1.

Manşonu kol üzerine, kalp seviyesinde
yerleştirin (tercihen sol kol).

Kalp seviyesindeŞuna basın: STARTdüğmesi2.

START düğmesine basın.

3.

Sıfır yanıp sönerken, beklenen sistolik
basıncınızdan yaklaşık 30 - 40 mmHg
yüksek bir sayı görünene kadar
START düğmesini basılı tutun.

4.

İstediğiniz sayıya erişildiğinde, ölçümübaşlatmak için START düğmesinibırakın. Önceki sayfada açıklandığı
gibi tansiyonunuzu ölçmeye devamedin.

İstenen sistolik
basınçta düğmeyi
bırakın

Düğmeyi basılı START
tutun

Ölçüm için önceki sayfaya
bakınDoğru Ölçümle İlgili Notlar
Rahat bir konumda oturun. Avuç içiniz yukarı bakacak şekilde vemanşonu kalbinizle aynı düzeyde tutarak kolunuzu masa üzerine
yerleştirin.
Ölçüme başlamadan önce beş ila on dakika kadar dinlenin.
Heyecanlıysanız veya duygusal baskı altındaysanız bu stres ölçüme
yansıyarak normal tansiyon değerinden daha yüksek (veya daha düşük)bir sonuç çıkmasına neden olabilir ve okunan nabız değeri genellikle
normalden fazla olur.
Bir kişinin tansiyonunu yapılan işe ve yenilenlere bağlı olarak süreklideğişir. İçtiğiniz şeyler, tansiyonunuzu çok güçlü ve hızlı bir şekilde
etkileyebilir.
Bu aygıt, ölçümlerini kalp atışına göre yapar. Çok zayıf veya düzensiz bir
kalp atışınız varsa aygıt, tansiyonunuzu ölçmede zorlanabilir.
Aygıt anormal bir durum algıladığında ölçmeyi durdurur ve bir hatasembolü görüntüler. Sembollerin açıklaması için bkz. sayfa 7.
Türkçe15

Bu tansiyon monitörü, yetişkinlerin kullanması için tasarlanmıştır. Bu
aygıtı çocuklar için kullanmadan önce doktorunuza danışın. Çocuklar, bu
aygıtı gözetimsiz kullanmamalıdır.
Otomatik tansiyon monitörünün performansı, aşırı sıcaklık veya nemden
ya da rakımdan etkilenebilir.

Bellek Verilerini Geri Çağırma
Not:

Bu aygıt son 60 ölçümü bellekte depolar.

1. Hiçbir şey görüntülenmediğinde,
depolanan verileri geri çağırmak
için _START_ düğmesini basılı
tutun.

Basılı
tutun

START

düğmesi
Ortalama sistolik
Ortalama diyastolik
Ortalama nabız

Tüm verilerin
ortalaması

2. Ortalama veri görüntülenirkendüğmeyi bırakın.

3. Veri numarası ve depolanan veriler
son ölçümden itibaren otomatikolarak görüntülenir.

Sistolik
Diyastolik

4. Tüm veriler görüntülendikten sonra
ekran otomatik olarak kapanır.

Not:

Nabız
En yeni veriler

Verileri geri çağırırken _START_düğmesine basarsanız aygıt
kapanır.

Sistolik
Diyastolik
Nabız
Son veriler (En eski)

Türkçe16

IHB/AFib Göstergesi Nedir?
Monitör, ölçümler sırasında düzensiz ritim algıladığında IHB/AFibgöstergesi, ölçüm değerleriyle birlikte ekranda görüntülenir.
Not:

Bu
IHB/AFib göstergesini sık sık görüyorsanız doktorunuzabaşvurmanızı öneririz.

AFib Nedir?
Kalp, oluşan elektrik sinyalleri nedeniyle kasılır ve tüm gövdeye kan
gönderir. Atriyumdaki elektrik sinyali karışırsa ve nabız aralıklarında
bozukluğa neden olursa atriyal fibrilasyon (AFib) oluşur. AFib, kanın
kalpte birikmesine neden olarak felç ve kalp krizini tetikleyen kan
pıhtılarına yol açar.

％IHB/AFib%IHB/AFib, algılanan IHB/AFib’in sıklığı olarak görüntülenir.
%IHB/AFib yalnızca fiziki hareket gibi gürültüleri değil düzensiz kalpatışını da algılayabilir. Bu yüzden, %IHB/AFib düzeyi yükseksedoktorunuza başvurmanızı öneririz.
%IHB/AFib =Bellekteki algılanan
IHB/AFib’lerin sayısı

x 100 [%]

Toplam sayı

%IHB/AFib ekranı: %IHB/AFib, ortalama değerlerle birlikte görüntülenir.
(Bkz. “2. Verileri Geri Çağırma“, “Çalışma Modu“ bölümü)
Bellek sayısı altı veya daha azsa %IHB/AFib
görüntülenmez.
Ortalama değer ekranı

Düzey 0
%IHB/AFib=0

Düzey 1Düzey 2Düzey 3
%IHB/AFib=1 - 9 %IHB/AFib=10 - 24 %IHB/AFib=25 - 100

Görüntülenmez

Türkçe17

Basınç Çubuğu Göstergesi
Gösterge ölçüm sırasında basıncın ilerleyişini izler.
Basınç veriliyor

Hava salınıyorŞişirme devam ediyorÖlçme devam ediyorŞişirme tamam

WHO Sınıflandırma GöstergesiÇubuk göstergesinin her bir segmenti,
sonraki sayfada tanımlanan WHO tansiyon
sınıflandırmasına karşılık gelir.
WHO Sınıflandırma GöstergesiŞiddetli yüksek tansiyon
Orta derecede yüksek tansiyon
Hafif yüksek tansiyon
Yüksek normal
NormalOptimumÖrnek:

Orta derecede yüksek tansiyon

Hafif yüksek tansiyon: Gösterge geçerli verilere göre WHOsınıflandırmasına karşılık gelen bir
segment görüntüler.
Yüksek normal

Tansiyon Hakkında
Tansiyon Nedir?
Tansiyon, arter duvarlarına kan tarafından uygulanan kuvvettir. Sistolik
basınç kalp kasıldığında oluşur. Diyastolik basınç kalp genişlediğinde
oluşur. Tansiyon, milimetre cıva cinsinden ölçülür (mmHg). Kişinin doğal
tansiyonu, kişi hâlâ dinlenirken ve yemek yemeden önce sabah ilk işolarak ölçülen temel basınç ile temsil edilir.

Türkçe18

Hipertansiyon Nedir ve Nasıl Kontrol Edilir?
Hipertansiyon, anormal şekilde yüksek arter tansiyonudur ve
ilgilenilmezse felç ve kalp krizi dahil birçok sağlık sorununa neden olabilir.
Hipertansiyon, yaşam tarzı değiştirilerek, stresten uzak durularak ve
doktor kontrolünde ilaç tedavisiyle kontrol edilebilir.
Hipertansiyonu önlemek veya kontrol altında tutmak için:
Sigara içmeyin
Düzenli egzersiz yapın
Tuz ve yağ alımınızı azaltın
Düzenli fiziki check-up yaptırın
Uygun ağırlığınızı koruyun

Tansiyon Neden Evde Ölçülmelidir?
Klinikte veya muayenehanede yapılan tansiyon ölçümü endişeye neden
olabilir ve okunan değerler evde ölçülenden 25 - 30 mmHg daha yüksek
çıkabilir. Evde ölçüm, tansiyon okunan değerlerinde dış etkileri azaltır,
doktorun okuduğu değerleri destekler ve daha doğru, tam bir tansiyon
geçmişi sağlar.

WHO Tansiyon Sınıflandırması

Tansiyon Değişimleri

Referans Materyal: Journal of Hypertension 1999, Cilt 17 No.2
mmHg
Derece 3 hipertansiyon (ciddi)
110
Derece 2 hipertansiyon (orta)
105
Diyastolik tansiyon

Dünya Sağlık Örgütü (WHO)tarafından belirlenmiş veçizelgede gösterildiği gibiyaşa bakılmaksızın yüksek
tansiyonu değerlendiren
standartlardır.

100
95
90
85
80

Derece 1 hipertansiyon (hafif)
Yüksek-normal
NormalTansiyonKişinin tansiyonu günlük ve
Optimum
120 130 140 150 160 170 180
mevsimsel olarak büyük
Cinsinden sistolik tansiyon
mmHgölçüde değişir. Gün
boyunca çeşitli koşullara bağlı olarak 30 ila 50 mmHg arasında değişebilir.
Tansiyonu yüksek kişilerde değişimler daha da fazla olabilir. Tansiyon,
genellikle çalışırken veya oynarken yükselir ve uyku sırasında en düşük
düzeyine düşer. Bu yüzden, tek bir ölçümün sonucu üzerine çok fazlaendişelenmeyin.
Normal tansiyonunuzu öğrenmek için bu
Gün içinde tipik dalgalanma(Her beş dakikada bir ölçüm)
kılavuzda
açıklananprosedürü
Sistolikkullanarak her gün aynı saatte ölçüm
yapın. Düzenli yapılan ölçümler dahakapsamlı tansiyon geçmişi sağlar.
Tansiyonunuzu kaydederken tarih ve
saati not ettiğinizden emin olun.
Uyku
Diyastolik
Tansiyon verilerinizi yorumlamak için
ÖS
ÖÖ
Saat
doktorunuza danışın.

Türkçe19

Sorun giderme
Sorun
Aygıt açık olsa
bile ekranda
hiçbir şey
görünmüyor.
Manşonşişmiyor.

Aygıt ölçüm
yapmıyor.
Okunandeğerler çok
yüksek veya
çok düşük.

Olası Neden
Piller bitmiştir.
Pil uçları doğru
konumda değildir.
Pil voltajı çok düşüktür.
(DÜŞÜK PİL işareti)
yanıp sönüyor. Piller
tamamen bittiyse işaret
görünmez.
Manşon düzgün şekilde
takılmamıştır.
Ölçüm sırasındakolunuzu veya
gövdenizi hareket
ettirmişsinizdir.

Önerilen Eylem
Tüm pilleri yenileriyle değiştirin.
Pilleri, negatif ve pozitif uçları pilbölmesinde gösterilen şekillerleeşleşecek şekilde yeniden
takın.
Tüm pilleri yenileriyle değiştirin.

Manşonu düzgün takın.
Ölçüm sırasında çok hareketsizve sessiz kaldığınızdan eminolun.

Rahat ve hareketsiz oturun.
Avuç içiniz yukarı bakacakManşon konumu doğru
şekilde ve manşonu kalbinizledeğil.
aynı düzeyde tutarak kolunuzu
masa üzerine yerleştirin.
Çok zayıf veya düzensiz bir kalpatışınız varsa aygıt
tansiyonunuzu ölçmede
zorlanabilir.
Değer, klinikte veya
“Tansiyon Neden EvdemuayenehanedeÖlçülmelidir?“ bölümüne
ölçülenden farklı.
başvurun.
Diğer
Pilleri çıkarın. Yerine geri takın
ve yeniden ölçmeyi deneyin.
Not: Yukarıda açıklanan eylemler sorunu çözmezse bayi ile temasa geçin.
Bu ürünün içini açmaya veya onarmaya çalışmayın; bunu yapmaya
kalkarsanız garantiniz geçersiz olur.

Türkçe20

BakımAygıtın içini açmayın. Hassas elektrikli bileşenler ve hasar görebilecek
karmaşık bir hava ünitesi kullanmaktadır. Sorun giderme talimatlarını
kullanarak sorunu gideremezseniz, bölgenizdeki yetkili bayiye veya
müşteri hizmetleri departmanımıza başvurun. A&D müşteri hizmetleri,yetkili bayilere teknik bilgi, yedek parça ve üniteler sağlar.
Aygıt uzun kullanım ömrü için tasarlanmış ve üretilmiştir. Ancak, aygıtındüzgün ve doğru çalışması için genellikle 2 yılda bir incelenmesi önerilir.
Lütfen bakım için bölgenizdeki yetkili bayiye ya da A&D’ye başvurun.

Teknik Veriler
TipÖlçüm yöntemi
Ölçüm aralığı

UA-767S
Osilometrik ölçüm
Basınç:0 - 299 mmHg
Sistolik basınç: 60 - 279 mmHg
Diyastolik basınç: 40 - 200 mmHg
Nabız:
40 – 180 atış / dakikaÖlçüm doğruluğu
Basınç:±3 mmHg
Nabız:%±5
Güç kaynağı
4 x 1,5V pil (R6P, LR6 veya AA) veya
AC adaptörü (TB-233C) (Dahil değildir)Ölçüm sayısı
Yakl. 700 defa
LR6 (alkalin piller)
Yaklaşık 200 defa R6P (manganez piller)
23 °C oda sıcaklığında 180 mmHg basınç ile.
Sınıflandırma
Dahili güçle çalışan ME ekipmanı (Pillerle
beslenmiştir) / Sınıf II (Adaptörle beslenmiştir)Sürekli çalışma modu
Klinik test
ISO81060-2：2013’e göre
Klinik doğrulama çalışmasında, diyastolik tansiyonun
tespit edilmesi için 85 gönüllü üzerinde K5
kullanılmıştır.
EMD
IEC 60601-1-2: 2014
BellekSon 60 ölçüm
Kullanım koşulları+10 ilâ +40 °C / %15 ilâ 85 BN / 800 ilâ 1.060 hPa
Nakliye / Saklama koşulları -20 ilâ +60 °C / %10 ilâ 95 BN / 700 ilâ 1.060 hPa
BoyutlarYakl. 140 [G] x 60 [Y] x 105 [D] mm
Ağırlık
Yakl. 245 g, piller hariçGiriş koruması
Aygıt:
IP20
Takılan kısımManşonTip BF

Türkçe21

Kullanım ömrüAygıt:
5 yıl (günde altı kez kullanıldığında)Manşon: 2 yıl (günde altı kez kullanıldığında)AC adaptörü: 5 yıl (günde altı defa kullanıldığında)Aksesuar AC adaptörü
Adaptör, aygıtı evdeki bir güç kaynağına bağlar. Satın almak için lütfen yerel
A&D bayinize başvurun. AC adaptörünün periyodik olarak incelenmesi veya
değiştirilmesi gerekir.
TB-233C
Satın almak için lütfen yerel A&D bayinizebaşvurun.
AC adaptörünün periyodik olarak incelenmesi veya
değiştirilmesi gerekir.
AC adaptörü üzerine basılmış semboller
Semboller
Fonksiyon / Anlamı
Yalnızca iç mekanda kullanım içindir
Sınıf II aygıtı
Termal sigorta
SigortaEC yönergesi aygıtı etiketi
EAC sertifika aygıtı etiketi
AC adaptörü fişinin polaritesi
Aksesuarlar ayrıca satılırManşon Katalog NumarasıManşon Boyutu
CUF-F-LA
Büyük yetişkin manşonu
CUF-IGeniş aralıklı manşon
CUF-F-AYetişkin manşonu
Kol boyutu: Pazı çevresi.
AC adaptörü
Not:Katalog Numarası
TB-233C

Kol Boyutu
31 cm - 45 cm
22 cm - 42 cm
22 cm - 32 cm
Fiş (Priz türü)
Tip C

Teknik özellikler önceden haber verilmeksizin değiştirilebilir.
IP sınıflandırması, IEC 60529’a uygun kapaklar tarafından sağlanan
koruma derecesidir. Aygıt, parmaklar gibi 12 mm ve üstü çaptaki katı
yabancı nesnelere karşı korunur. Bu aygıt, suya karşı dayanıklıdeğildir.

Türkçe22‫اﻟﻣﺣﺗوﯾﺎت‬‫اﻟﻌﻣﻼء اﻷﻋزاء ‪2 ........................................................................................‬‬‫ﻣﻼﺣظﺎت أوﻟﯾّﺔ ‪2 ........................................................................................‬‬‫اﻟﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ ‪2 .....................................................................................‬‬‫اﻟﺗﻌرﯾف ﺑﺄﺟزاء اﻟﺟﮭﺎز ‪5 ...............................................................................‬‬‫اﻟرﻣوز ‪6 . ................................................................................................‬‬‫ﻧﻣط اﻟﺗﺷﻐﯾل‪8 ............................................................................................‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس ‪9 ..................................................................................‬‬‫ﺗرﻛﯾب ‪ /‬ﺗﻐﯾﯾر اﻟﺑطﺎرﯾﺎت ‪9 .......................................................................‬‬‫ﺗوﺻﯾل ﺧرطوم اﻟﮭواء ‪10 ........................................................................‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪10 ............................................................‬‬‫ﺗوﺻﯾل ﻗﺎﺑس ّ‬‫اﺧﺗﯾﺎر اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟﻠرﺑﺎط ‪11 ................................................................‬‬‫ﻛﯾﻔﯾﺔ وﺿﻊ رﺑﺎط اﻟذراع ‪11 ......................................................................‬‬‫ﻛﯾﻔﯾﺔ إﺟراء ﻋﻣﻠﯾﺎت ﻗﯾﺎس دﻗﯾﻘﺔ ‪13 ..............................................................‬‬‫اﻟﻘﯾﺎس ‪13 ...........................................................................................‬‬‫ﺑﻌد اﻟﻘﯾﺎس ‪13 ......................................................................................‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ‪14 .......................................................................................‬‬‫اﻟﻘﯾﺎس اﻟﻌﺎدي‪14 ...................................................................................‬‬‫إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ ‪15 .. ................................‬‬‫ﻣﻼﺣظﺎت ﻟﻘﯾﺎس دﻗﯾﻖ ‪15 .........................................................................‬‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة ‪16 ........................................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣؤﺷر ‪IHB/AFib‬؟ ‪17 . ................................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑـ ‪AFib‬؟ ‪17 ..............................................................................‬‬‫‪17 ..................................................................................... ％IHB/AFib‬‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط ‪18 ................................................................................‬‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ‪18 .............................................................‬‬‫ﻧُﺑذة ﻋن ﻣﺳﺗوى ﺿﻐط اﻟدم ‪18 .........................................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣﺳﺗوى ﺿﻐط اﻟدم؟ ‪18 .............................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎرﺗﻔﺎع ﺿﻐط اﻟدم وﻛﯾف ﯾﻣﻛن اﻟﺗﺣﻛم ﻓﯾﮫ؟ ‪19 .... ................................‬‬‫ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟ ‪19 ................................................‬‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم ‪19 .........................................‬‬‫اﻻﺧﺗﻼﻓﺎت ﻓﻲ ﺿﻐط اﻟدم ‪19 ..... ................................................................‬‬‫اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ ‪20 ......................................................................‬‬‫اﻟﺻﯾﺎﻧﺔ ‪21 ...............................................................................................‬‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻔﻧﯾﺔ ‪21 ........................................................................................‬‬‫اﻟﻌﺮﺑﯿﺔ ‪1‬‬‫اﻟﻌﻣﻼء اﻷﻋزاء‬‫ﺗﮭﺎﻧﯾﻧﺎ ﻟﻛم ﻋﻠﻰ ﺷراء أﺣدث ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻣن ‪ ،A&D‬اﻟذي ﯾﻌﺗﺑر أﺣد اﻷﺟﮭزة اﻷﻛﺛر‬‫ﺻ ّﻣم اﻟﺟﮭﺎز ﻟﯾوﻓر ﺳﮭوﻟﺔ ﻓﻲ اﻻﺳﺗﺧدام ودﻗﺔً ﻓﻲ اﻟﻘﯾﺎس‪ ،‬ﺣﯾث ﯾﺳ ّﮭل ﻋﻠﯾك‬‫ﺗطو ًرا ﻓﻲ اﻟوﻗت اﻟﺣﺎﻟﻲ‪ُ .‬‬‫ّ‬‫اﻟﻘﯾﺎس اﻟﯾوﻣﻲ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﻧُوﺻﯾﻛم ﺑﻘراءة ھذا اﻟدﻟﯾل ﺑﻌﻧﺎﯾﺔ ﻗﺑل اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻠﻣرة اﻷوﻟﻰ‪.‬‬‫ﻣﻼﺣظﺎت أوﻟﯾّﺔ‬‫ﯾﺗطﺎﺑﻖ ھذا اﻟﺟﮭﺎز ﻣﻊ اﻟﻣواﺻﻔﺎت اﻷوروﺑﯾﺔ ‪ 93/42 EEC‬اﻟﺧﺎﺻﺔ ﺑﺎﻷﺟﮭزة اﻟطﺑﯾﺔ‪ .‬وﯾﺷﯾر رﻣز‬‫إﻟﻰ دﻗﺔ اﻟﺗطﺎﺑﻖ‪ :0123) .‬اﻟرﻗم اﻟﻣرﺟﻌﻲ ﻟﻠﮭﯾﺋﺔ اﻟﻣﻌﻧﯾﺔ اﻟﺗﻲ ﺗم إﺧطﺎرھﺎ(‬‫ﺻ ّﻣم ھذا اﻟﺟﮭﺎز ﻟﻼﺳﺗﺧدام ﻣﻊ اﻟﻛﺑﺎر دون اﻷطﻔﺎل اﻟﺻﻐﺎر وﺣدﯾﺛﻲ اﻟوﻻدة‪.‬‬‫ُ‬‫ﻣﻛﺎن اﻻﺳﺗﺧدام‪ .‬ھذا اﻟﺟﮭﺎز ُﻣﺧﺻص ﻟﻼﺳﺗﻌﻣﺎل ﻷﻏراض اﻟرﻋﺎﯾﺔ اﻟﺻﺣﯾﺔ اﻟﻣﻧزﻟﯾﺔ‪.‬‬‫ﺻﻣم ھذا اﻟﺟﮭﺎز ﻟﻘﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم وﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟﻺﻧﺳﺎن ﻷﻏراض اﻟﺗﺷﺧﯾص‪.‬‬‫ُ‬‫اﻟﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ‬‫ﻛوﻧﺎت دﻗﯾﻘﺔ؛ ﻟذا ﺗﺟﻧّب ﻣطﻠﻘًﺎ ﺗﻌرﯾﺿﮫ ﻟدرﺟﺎت اﻟﺣرارة واﻟرطوﺑﺔ اﻟﻣرﺗﻔﻌﺔ‬‫ُ‬‫ﺻﻧّﻊ ھذا اﻟﺟﮭﺎز ﺑﺎﺳﺗﺧدام ُﻣ ّ‬‫وأﺷﻌﺔ اﻟﺷﻣس اﻟﻣﺑﺎﺷرة واﻟﺻدﻣﺎت واﻟﻐﺑﺎر‪.‬‬‫ّ‬‫ﻧظف اﻟﺟﮭﺎز واﻟرﺑﺎط ﺑﺎﺳﺗﺧدام ﻗطﻌﺔ ﻗﻣﺎش ﺟﺎﻓﺔ وﻧﺎﻋﻣﺔ أو ﻗطﻌﺔ ﻗﻣﺎش ﻣﺑﻠﻠﺔ ﺑﺎﻟﻣﺎء وﻣﻧظف ﻣﻌﺗدل‪.‬‬‫ﺗﺟﻧب ﻣطﻠﻘًﺎ اﺳﺗﺧدام اﻟﻛﺣول أو اﻟﺑﻧزﯾن أو اﻟﺛﻧر أو ﻏﯾرھﺎ ﻣن اﻟﻣواد اﻟﻛﯾﻣﯾﺎﺋﯾﺔ اﻟﻘﺎﺳﯾﺔ ﻣن أﺟل ﺗﻧظﯾف‬‫اﻟﺟﮭﺎز أو اﻟرﺑﺎط‪.‬‬‫ﻲ اﻟرﺑﺎط ﺑﺷدة أو ﺗﺧزﯾن اﻟﺧرطوم ﺑﺛﻧﯾﮫ ﺑﺷدة ﻟﻔﺗرات طوﯾﻠﺔ؛ ﯾؤدي ھذا اﻟﺗﺻرف إﻟﻰ ﻗﺻر ﻣدة‬‫ﺗﺟﻧب ط ّ‬‫اﻟﻣﻛوﻧﺎت‪.‬‬‫اﺳﺗﺧدام ﺗﻠك‬‫ّ‬‫ﺗو ﱠخ اﻟﺣذر ﻟﺗﻔﺎدي أي ﺣوادث اﺧﺗﻧﺎق ﻋن طرﯾﻖ اﻟﺧطﺄ ﻟﻸطﻔﺎل واﻟرﺿﻊ ﻋن طرﯾﻖ ھذا اﻟﺧرطوم أو‬‫اﻟﺳﻠك‪.‬‬‫ﻧظرا‬‫ﻻ ﺗﻘم ﺑﺛﻧﻲ ﺧرطوم اﻟﮭواء أﺛﻧﺎء إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻟﺗﻌرض ﻹﺻﺎﺑﺔ‬‫ً‬‫ﻟﻠﺿﻐط اﻟﻣﺳﺗﻣر ﻋﻠﻰ اﻟرﺑﺎط‪.‬‬‫ﺗﻌرض اﻟﺟﮭﺎز واﻟرﺑﺎط ﻟﻠﺑﻠل ﺑﻔﻌل اﻷﻣطﺎر واﻟﻌرق واﻟﻣﯾﺎه‪.‬‬‫اﻟﺟﮭﺎز واﻟرﺑﺎط ﻏﯾر ﻣﻘﺎوﻣﯾن ﻟﻠﻣﺎء‪ .‬ﺗﺟﻧﱠب ّ‬‫ﯾﻣﻛن أن ﺗﺗﻌرض ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻠﺗﺷوﯾش إذا ﺟرى اﺳﺗﺧدام اﻟﺟﮭﺎز ﺑﺟوار أﺟﮭزة اﻟﺗﻠﻔزﯾون أو‬‫اﻟﻣﯾﻛرووﯾف أو اﻟﮭواﺗف اﻟﻣﺣﻣوﻟﺔ أو اﻷﺟﮭزة اﻟﺗﻲ ﯾﻧﺑﻌث ﻣﻧﮭﺎ اﻷﺷﻌﺔ اﻟﺳﯾﻧﯾﺔ أو اﻷﺧرى ذات اﻟﻣﺟﺎﻻت‬‫اﻟﻛﮭرﺑﺎﺋﯾﺔ اﻟﻘوﯾﺔ‪.‬‬‫أﺟﮭزة اﻻﺗﺻﺎﻻت اﻟﻼﺳﻠﻛﯾﺔ‪ ،‬ﻣﺛل أﺟﮭزة اﻻﺗﺻﺎل اﻟﺷﺑﻛﻲ اﻟﻣﻧزﻟﯾﺔ‪ ،‬واﻟﮭواﺗف اﻟﻣﺣﻣوﻟﺔ‪ ،‬واﻟﮭواﺗف‬‫اﻟﻼﺳﻠﻛﯾﺔ واﻟﻣﺣطﺎت اﻟﻘﺎﻋدﯾﺔ ﻟﮭﺎ‪ ،‬واﻷﺟﮭزة اﻟﻼﺳﻠﻛﯾﺔ اﻟﺗﻲ ﺗﻌﻣل ﺑﻣوﺟﺎت اﻟرادﯾو‪ ،‬ﺟﻣﯾﻌﮭﺎ ﯾﻣﻛن أن ﺗؤﺛر‬‫ﻋﻠﻰ ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ .‬وﻟذﻟك‪ ،‬ﯾﺟب اﻟﺑﻘﺎء ﻋﻠﻰ ﻣﺳﺎﻓﺔ ﻻ ﺗﻘل ﻋن ‪ 30‬ﺳم ﻣن ھذه اﻷﺟﮭزة‪.‬‬‫ﻋﻧد إﻋﺎدة اﺳﺗﺧدام اﻟﺟﮭﺎز‪ ،‬ﺗﺄﻛد ﻣن أﻧﮫ ﻧظﯾف‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪2‬‬‫اﻷﺟﮭزة واﻷﺟزاء واﻟﺑطﺎرﯾﺎت اﻟﻣﺳﺗﻌﻣﻠﺔ ﻻ ﺗُﻌﺎﻣل ﻣﻌﺎﻣﻠﺔ اﻟﻧﻔﺎﯾﺎت اﻟﻣﻧزﻟﯾﺔ اﻟﻌﺎدﯾﺔ‪ ،‬وﯾﺟب اﻟﺗﺧﻠص ﻣﻧﮭﺎ‬‫وﻓﻘًﺎ ﻟﻠواﺋﺢ اﻟﻣﺣﻠﯾﺔ اﻟﻣﻌﻣول ﺑﮭﺎ‪.‬‬‫اﻟﻣﺣول ﻣن اﻟﻣﺧرج اﻟﻛﮭرﺑﺎﺋﻲ ﻋﻧد‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪ ،‬ﺗﺄﻛد ﻣن ﺳﮭوﻟﺔ إﻣﻛﺎﻧﯾﺔ إزاﻟﺔ‬‫ﻋﻧد اﺳﺗﺧدام‬‫ّ‬‫ّ‬‫اﻟﺿرورة‪.‬‬‫ﻻ ﺗﻌدّل ﻓﻲ اﻟﺟﮭﺎز‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ وﻗوع ﺣوادث أو ﺗﻠف اﻟﺟﮭﺎز‪.‬‬‫ﯾﺗطﻠب ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬اﻟﺿﻐط ﻋﻠﻰ اﻟذراع ﻣن ﺧﻼل اﻟرﺑﺎط ﺑﻘوة ﺗﻛﻔﻲ ﻹﯾﻘﺎف ﺗدﻓﻖ اﻟدم ﻣؤﻗﺗًﺎ‬‫ﻓﻲ ﺷرﯾﺎن اﻟذراع‪ .‬ﯾﻣﻛن أن ﯾُﺳﺑّب ذﻟك ﻓﻲ اﻟﺷﻌور ﺑﺄﻟم أو ﺗﻧﻣﯾل أو ظﮭور ﻋﻼﻣﺔ ﺣﻣراء ﻣؤﻗﺗﺔ ﺑﺎﻟذراع‪.‬‬‫ﺗظﮭر ھذه اﻟﻌﻼﻣﺎت ﻻ ﺳﯾﻣﺎ ﻋﻧد ﺗﻛرار ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس ﺑﺷﻛل ﻣﺗﺗﺎﺑﻊ‪ .‬ﺳﯾﺧﺗﻔﻲ أي أﻟم أو ﺗﻧﻣﯾل أو ﻋﻼﻣﺎت‬‫ﺣﻣراء ﺑﻣرور اﻟوﻗت‪.‬‬‫ﯾﻣﻛن أن ﯾﺗﺳﺑب ﺗﻛرار ﻋﻣﻠﯾﺎت ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﺑﺷﻛل ﻣﺗﺗﺎﺑﻊ‪ ،‬ﻓﻲ ﺣدوث ﻣﺷﻛﻠﺔ ﺻﺣﯾﺔ ﺑﺳﺑب‬‫ﺗﻛرار اﻟﺿﻐط ﻋﻠﻰ ﺗدﻓﻖ اﻟدم‪ .‬ﺗﺄﻛد ﻣن أن ﻋﻣل اﻟﺟﮭﺎز ﻻ ﯾؤدي إﻟﻰ ﺿﻌف طوﯾل اﻷﻣد ﻓﻲ اﻟدورة‬‫اﻟدﻣوﯾﺔ‪ ،‬ﻋﻧد اﺳﺗﺧدام اﻟﺟﮭﺎز ﺑﺎﺳﺗﻣرار‪.‬‬‫ت ﻋﻣﻠﯾﺔ اﺳﺗﺋﺻﺎل اﻟﺛدي‪ ،‬ﻓﺎﺳﺗﺷﯾري اﻟطﺑﯾب ﻗﺑل اﺳﺗﺧدام اﻟﺟﮭﺎز‪.‬‬‫إذا ﻛﻧت ﻗد أﺟرﯾ ِ‬‫ﻻ ﺗﺳﻣﺢ ﻟﻸطﻔﺎل ﺑﺎﺳﺗﺧدام اﻟﺟﮭﺎز ﻣن ﺗﻠﻘﺎء أﻧﻔﺳﮭم واﺣرص ﻋﻠﻰ أن ﯾﻛون اﺳﺗﺧدام اﻟﺟﮭﺎز ﻓﻲ ﻣﻛﺎن ﺑﻌﯾد‬‫ﻋن ﻣﺗﻧﺎول اﻷطﻔﺎل‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ وﻗوع ﺣوادث أو ﺗﻠف‪.‬‬‫ﺛﻣﺔ أﺟزاء ﺻﻐﯾرة ﻓﻲ اﻟﺟﮭﺎز ﯾﻣﻛن أن ﺗﺳﺑب اﻻﺧﺗﻧﺎق ﻟﻸطﻔﺎل ﻓﻲ ﺣﺎل اﺑﺗﻼﻋﮭﺎ ﻋن طرﯾﻖ اﻟﺧطﺄ‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻋﻧد ﻋدم اﺳﺗﺧداﻣﮫ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﻓﺻل ّ‬‫ﯾﻌرض اﺳﺗﺧدام ﻣﻠﺣﻘﺎت ﻻ ﺗرد ﺗﻔﺻﯾﻼً ﻓﻲ ھذا اﻟدﻟﯾل ﺳﻼﻣﺗك ﻟﻠﺧطر‪.‬‬‫ﻗد ّ‬‫إذا ﺗﺳﺑﺑت اﻟﺑطﺎرﯾﺔ ﻓﻲ ﺣدوث داﺋرة ﻗﺻر‪ ،‬ﻓﯾﻣﻛن أن ﺗﺻﺑﺢ ﺳﺎﺧﻧﺔ وﺗﺗﺳﺑب ﻓﻲ ﺣروق ﺑﺎﻟﺟﻠد‪.‬‬‫اﺗرك اﻟﺟﮭﺎز ﯾﺗﻛﯾّف ﻋﻠﻰ اﻟﺑﯾﺋﺔ اﻟﻣﺣﯾطﺔ ﻗﺑل اﻻﺳﺗﺧدام )ﻣدة ﺳﺎﻋﺔ واﺣدة ﺗﻘرﯾﺑًﺎ(‪.‬‬‫ﻟم ﺗُﺟر اﺧﺗﺑﺎرات ﺳرﯾرﯾﺔ ﻋﻠﻰ اﻷطﻔﺎل ﺣدﯾﺛﻲ اﻟوﻻدة واﻟﻧﺳﺎء اﻟﺣواﻣل‪ .‬ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻋﻠﻰ اﻷطﻔﺎل‬‫ﺣدﯾﺛﻲ اﻟوﻻدة أو اﻟﻧﺳﺎء اﻟﺣواﻣل‪.‬‬‫اﻟﺗﻌرض‬‫ﻻ ﺗﻠﻣس اﻟﺑطﺎرﯾﺎت وﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر واﻟﻣرﯾض ﺳوﯾًﺎ ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ‪ .‬ﻗد ﯾؤدي ذﻟك إﻟﻰ‬‫ّ‬‫ﻟﺻدﻣﺔ ﻛﮭرﺑﺎﺋﯾﺔ‪.‬‬‫ﻻ ﺗﺑدأ ﻓﻲ ﻧ ْﻔﺦ اﻟﮭواء دون ﻟف اﻟرﺑﺎط ﺣول اﻟﻌﺿد‪.‬‬‫ﻣواﻧﻊ اﻻﺳﺗﻌﻣﺎل‬‫ﻓﯾﻣﺎ ﯾﻠﻲ اﺳﺗﻌراض ﻟﻠﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ ﻣن أﺟل اﺳﺗﺧدام ﻣﻼﺋم ﻟﻠﺟﮭﺎز‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﯾﺗﺻل ﺑﮫ ﺟﮭﺎز طﺑﻲ ﻛﮭرﺑﺎﺋﻲ آﺧر‪ .‬ﻗد ﻻ ﯾﻌﻣل اﻟﺟﮭﺎز ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﯾﺟب ﻋﻠﻰ اﻷﺷﺧﺎص اﻟذﯾن ﯾﻌﺎﻧون ﻣن ﻧﻘص ﺷدﯾد ﻓﻲ اﻟدورة اﻟدﻣوﯾﺔ ﻓﻲ اﻟذراع اﺳﺗﺷﺎرة اﻟطﺑﯾب ﻗﺑل‬‫اﻟﺗﻌرض ﻟﻠﻣﺷﻛﻼت اﻟطﺑﯾﺔ‪.‬‬‫اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﺗﻔﺎدي‬‫ّ‬‫ﻻ ﺗﺷ ّﺧص ﺑﻧﻔﺳك ﻧﺗﺎﺋﺞ اﻟﻘﯾﺎس وﺗﺑدأ ﻓﻲ ﺗﻧﺎول اﻟﻌﻼج ﻣن ﺗﻠﻘﺎء ﻧﻔﺳك‪ .‬اﺳﺗﺷر اﻟطﺑﯾب داﺋ ًﻣﺎ ﻣن أﺟل ﺗﻘﯾﯾم‬‫اﻟﻧﺗﺎﺋﺞ واﻟﻌﻼج‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﺑﮫ ﺟرح ﻟم ﯾﻠﺗﺋم ﺑﻌد‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﯾﺗﻠﻘﻰ اﻟﺗﻧﻘﯾط اﻟورﯾدي أو ﻧﻘل اﻟدم‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻟﺗﻌرض‬‫ﻟﻺﺻﺎﺑﺔ أو اﻟﺿرر‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪3‬‬‫ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻓﻲ أﻣﺎﻛن وﺟود اﻟﻐﺎزات اﻟﻘﺎﺑﻠﺔ ﻟﻼﺷﺗﻌﺎل‪ ،‬ﻣﺛل ﻏﺎزات اﻟﺗﺧدﯾر‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك‬‫ﻓﻲ ﺣدوث اﻧﻔﺟﺎر‪.‬‬‫ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻓﻲ أﻣﺎﻛن ﺗرﻛﯾز اﻷﻛﺳﺟﯾن ﻓﯾﮭﺎ ﻣرﺗﻔﻊ‪ ،‬ﻣﺛل ﻏرﻓﺔ اﻷﻛﺳﺟﯾن ﻋﺎﻟﯾﺔ اﻟﺿﻐط أو ﺧﯾﻣﺔ‬‫اﻷﻛﺳﺟﯾن‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻧدﻻع ﺣرﯾﻖ أو ﺣدوث اﻧﻔﺟﺎر‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪4‬‬‫اﻟﺗﻌرﯾف ﺑﺄﺟزاء اﻟﺟﮭﺎز‬‫اﻟﺷﺎﺷﺔ‬‫ﺻل اﻟﮭواء‬‫ﻗﺎﺑس ﻣو ّ‬‫ﻗﺎﺑس ﻣﺣ ّول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫اﻟزر ‪START‬‬‫ﻣﻘﺑس اﻟﮭواء‬‫ﺧرطوم اﻟﮭواء‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫رﻣز ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫رﻣز اﻟﻣؤﺷر‬‫رﺑﺎط اﻟذراع‬‫ﻏطﺎء اﻟﺑطﺎرﯾﺎت‬‫ﺣﺟرة اﻟﺑطﺎرﯾﺎت‬‫ﺑطﺎرﯾﺎت ‪ 1.5‬ﻓوﻟت‬‫)‪ R6P‬أو ‪ LR6‬أو ‪(AA‬‬‫اﻟﺸﺎﺷﺔ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫اﻟذاﻛرة‬‫رﻣز ‪IHB/AFib‬‬‫اﻧﻘﺑﺎﺿﻲ‬‫اﻟﺿﻐط‬‫‪% IHB/AFib‬‬‫رﻣز ﺧطﺄ ﻓﻲ ﺗﺛﺑﯾت اﻟرﺑﺎط‬‫اﻟﺿﻐط‬‫اﻻﻧﺑﺳﺎطﻲ‬‫رﻣز ﺧطﺄ اﻟﺣرﻛﺔ‬‫رﻣز اﻟﻘﻠب‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‬‫ﺳرﻋﺔ اﻟﻧﺑض‬‫ﻣؤﺷر اﻟﺑطﺎرﯾﺔ‬‫اﻟﻌرﺑﯾﺔ ‪5‬‬

‫اﻟرﻣوز‬‫اﻟرﻣوز اﻟﺗﻲ ﯾﻣﻛن طﺑﺎﻋﺗﮭﺎ ﻋﻠﻰ ﻋﻠﺑﺔ اﻟﺟﮭﺎز‬‫اﻟرﻣوز‬‫‪SYS‬‬‫‪DIA‬‬‫‪PUL‬‬‫‪IP‬‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫وﺿﻊ اﻻﺳﺗﻌداد وﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫ﺿﻐط اﻟدم اﻻﻧﻘﺑﺎﺿﻲ ﺑـ ‪mmHg‬‬‫ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ ﺑـ ‪mmHg‬‬‫ﻋدد اﻟﺿرﺑﺎت ﻓﻲ اﻟدﻗﯾﻘﺔ‬‫دﻟﯾل ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت‬‫ﺗﯾﺎر ﻣﺑﺎﺷر‬‫اﻟﻧوع ‪ :BF‬ﺗم ﺗﺻﻣﯾم اﻟﺟﮭﺎز‪ ،‬واﻟرﺑﺎط‪ ،‬واﻷﻧﺑوب ﻟﺗوﻓﯾر ﺣﻣﺎﯾﺔ ﺧﺎﺻﺔ ﻣن اﻟﺻدﻣﺎت‬‫اﻟﻛﮭرﺑﺎﺋﯾﺔ‪.‬‬‫ﻣﻠﺻﻖ اﻟﺗوﺟﯾﮫ ‪ EC‬ﻟﻸﺟﮭزة اﻟطﺑﯾﺔ‬‫ﻣﻣﺛل اﻻﺗﺣﺎد اﻷوروﺑﻲ‬‫اﻟﺟﮭﺔ اﻟﻣﺻﻧّﻌﺔ‬‫ﺗﺎرﯾﺦ اﻟﺗﺻﻧﯾﻊ‬‫رﻣز اﻟﺣﻣﺎﯾﺔ اﻟدوﻟﯾﺔ‬‫رﻣز اﻟدﻟﯾل اﻟﺗوﺟﯾﮭﻲ ﻟﻧﻔﺎﯾﺎت اﻟﻣﻌدات اﻟﻛﮭرﺑﺎﺋﯾﺔ واﻹﻟﻛﺗروﻧﯾﺔ‬‫‪SN‬‬‫اﻟرﻗم اﻟﺗﺳﻠﺳﻠﻲ‬‫راﺟﻊ دﻟﯾل‪/‬ﻛﺗﯾّب اﻹرﺷﺎدات‬‫ﻗطﺑﯾﺔ ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫ﺣﺎﻓظ ﻋﻠﯾﮫ ﺟﺎﻓًﺎ‬‫اﻟرﻣوز اﻟﺗﻲ ﺗظﮭر ﻋﻠﻰ اﻟﺷﺎﺷﺔ‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﯾظﮭر ﺣﯾﻧﻣﺎ ﯾﻛون إﺟراء اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬ﯾوﻣض ﻋﻧد إﺟراء اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬ﯾظل ﺛﺎﺑﺗًﺎ‬‫ﻗدر اﻟﻣﺳﺗطﺎع‪.‬‬‫اﻟﻛﺷف ﻋن اﻟﻧﺑض‪.‬‬‫رﻣز ‪IHB/AFib‬‬‫ﯾظﮭر ﻋﻧد اﻟﻛﺷف ﻋن ﻋدم اﻧﺗظﺎم ﻓﻲ ﺿرﺑﺎت اﻟﻘﻠب‪.‬‬‫ﯾﻣﻛن أن ﯾُﺿﻲء ﻋﻧد اﻟﻛﺷف ﻋن وﺟود اھﺗزاز ﺑﺳﯾط‬‫ﻟﻠﻐﺎﯾﺔ‪ ،‬ﻣﺛل اﻻھﺗزاز أو اﻻرﺗﻌﺎش‪.‬‬‫ﯾﻣﻛن أن ﯾؤدي إﻟﻰ إظﮭﺎر ﻗﯾﻣﺔ‬‫ﺧطﺄ‪.‬‬‫ﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة‬‫ﺗﺣرك اﻟﺟﺳم أو اﻟذراع أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ﯾظﮭر ﻋﻧد ّ‬‫ُ‬‫ً‬‫أﺧرى‪ .‬ﻛن ﺛﺎﺑﺗﺎ أﺛﻧﺎء إﺟراء‬‫اﻟﻘﯾﺎس‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪6‬‬

‫اﻟرﻣوز‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﯾﻣﻛن أن ﯾؤدي إﻟﻰ إظﮭﺎر ﻗﯾﻣﺔ ﺧطﺄ‪.‬‬‫ﯾظﮭر أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس ﻓﻲ ﺣﺎل ﻋدم ﻟف اﻟرﺑﺎط‬‫ﺿﻊ اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ وﺣﺎول‬‫ﺑﺈﺣﻛﺎم ﺣول اﻟذراع‬‫إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ﻣؤﺷر إﻟﻰ ﻣﻌدل ‪ IHB/AFib‬اﻟ ُﻣﻛﺗﺷف ﻓﻲ‬‫اﻟذاﻛرة‬‫‪= %IHB/AFib‬‬‫ﻋدد ﺣﺎﻻت ‪IHB/AFib‬‬‫اﻟ ُﻣﻛﺗﺷﻔﺔ ﻓﻲ اﻟذاﻛرة‬‫‪[%] 100 X‬‬‫اﻟرﻗم اﻟﻛﻠﻲ‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس اﻟﺳﺎﺑﻘﺔ اﻟ ُﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‪.‬‬‫ﻣﺗوﺳط اﻟﺑﯾﺎﻧﺎت )ﻣﺗوﺳط ﻋدد اﻟﻘﯾﺎﺳﺎت اﻟﻣﺧزﻧﺔ(‬‫اﻟﺑطﺎرﯾﺔ ﻣﻣﺗﻠﺋﺔ‬‫ﻣؤﺷر طﺎﻗﺔ اﻟﺑطﺎرﯾﺔ ﻓﻲ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ‬‫اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ ﻋﻧدﻣﺎ ﯾوﻣض‪.‬‬‫اﻟﺗﺣرك ﻓﻲ أﺛﻧﺎء‬‫ﻧظرا إﻟﻰ‬‫ّ‬‫ﺿﻐط اﻟدم ﻏﯾر ﻣﺳﺗﻘر ً‬‫إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﻗﯾم اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ واﻻﻧﺑﺳﺎطﻲ ﺿﻣن ﻧطﺎق‬‫‪ mmHg 10‬ﻣن ﻛﻠﯾﮭﻣﺎ‪.‬‬‫ﻗﯾﻣﺔ اﻟﺿﻐط ﻻ ﺗرﺗﻔﻊ أﺛﻧﺎء ﻧ ْﻔﺦ اﻟﮭواء‬‫ﻟم ﯾُﻠف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫‪.PUL DISPLAY ERROR‬‬‫ﻟم ﯾﺗم اﻛﺗﺷﺎف ﺿرﺑﺎت اﻟﻘﻠب ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﺧطﺄ داﺧﻠﻲ ﻓﻲ ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫اﻟﻌرﺑﯾﺔ ‪7‬‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‬‫ﻋﻧدﻣﺎ ﯾوﻣض اﻟرﻣز‪.‬‬‫ﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ُﻛن ﺛﺎﺑﺗًﺎ ﺗﻣﺎ ًﻣﺎ أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪ ،‬وﺣﺎول‬‫إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت واﺿﻐط ﻋﻠﻰ اﻟزر‬‫‪ ، START‬ﺛ ّم أ ِﻋد ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت‬‫ﻣرة أﺧرى‪ .‬اﺗﺻل ﺑﺎﻟوﻛﯾل إذا اﺳﺗﻣر‬‫ظﮭور اﻟﺧطﺄ‪.‬‬‫ﻧﻣط اﻟﺗﺷﻐﯾل‬‫‪ .1‬اﻟﻘﯾﺎس اﻟﻌﺎدي‬‫اﺿﻐط ﻋﻠﻰ اﻟزر ‪ . START‬ﺗﺑدأ ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم وﺗُﺳ ّﺟل اﻟﺑﯾﺎﻧﺎت ﻓﻲ اﻟذاﻛرة‪ .‬ﯾﻣﻛن‬‫ﻟﻠﺟﮭﺎز ﺣﻔظ آﺧر ‪ 60‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻓﻲ اﻟذاﻛرة‪.‬‬‫‪ .2‬اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت‬‫ﻓﻲ وﺿﻊ اﻻﺳﺗﻌداد‪،‬‬‫اﺿﻐط ﺑﺎﺳﺗﻣرار‬‫ﻋﻠﻰ اﻟزر‬‫‪START‬‬‫ﻋﻧد ﻋدم ظﮭور ﺷﻲء‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ اﻟزر ‪. START‬‬‫ﺣرر اﻟزر ﻋﻧد ظﮭور ﻣﺗوﺳط اﻟﻘﯾﺎﺳﺎت اﻟﺳﺎﺑﻘﺔ اﻟﻣﺧزﻧﺔ‪.‬‬‫ّ‬‫ﺗُﻌرض رﻗم اﻟﻘﯾﺎﺳﺎت واﻟﻘﯾﺎﺳﺎت اﻟ ُﻣ ّ‬‫ﺧزﻧﺔ ﺗﻠﻘﺎﺋﯾًﺎ ﺑﺎﻟﺗرﺗﯾب وذﻟك ﻣن‬‫آﺧر ﻣرة ﻗﯾﺎس‪.‬‬‫ﻟﺗﻔﺎﺻﯾل ﻋن اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت‪ ،‬راﺟﻊ "اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن‬‫اﻟذاﻛرة"‪.‬‬‫‪ .3‬ﻣﺳﺢ ﻛل اﻟﺑﯾﺎﻧﺎت اﻟ ُﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‬‫ﻋﻧد إﻏﻼق اﻟﺟﮭﺎز‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ‬‫"‪.‬‬‫زر ‪ START‬ﺣﺗﻰ ﺗظﮭر "‬‫" ﻟﺣذف اﻟﺑﯾﺎﻧﺎت‪.‬‬‫اﺧﺗر "‬
‫ﺗُﺣذف اﻟﺑﯾﺎﻧﺎت ﻋﻧدﻣﺎ ﺗوﻣض‬‫ﯾﻧطﻔﺊ اﻟﺟﮭﺎز ﺗﻠﻘﺎﺋﯾًﺎ‪.‬‬‫‪START‬‬‫‪ .4‬إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ‬‫راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪ 15‬ﻟﻠﺗﻌرف ﻋﻠﻰ إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣطﻠوب‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪8‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﻟﺧطوة ‪1‬‬‫ﺗرﻛﯾب ‪ /‬ﺗﻐﯾﯾر اﻟﺑطﺎرﯾﺎت‬‫اﻟﺧطوة ‪3‬‬‫اﻟﺧطوة ‪1‬‬‫‪ .1‬اﻧزع ﻏطﺎء اﻟﺑطﺎرﯾﺎت‪.‬‬‫‪ .2‬اﻧزع اﻟﺑطﺎرﯾﺎت اﻟﻣﺳﺗﺧدﻣﺔ وأدﺧل اﻷﺧرى‬‫اﻟﺟدﯾدة ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت ﻛﻣﺎ ھو ﻣوﺿّﺢ‪ ،‬ﻣﻊ‬‫ﻣراﻋﺎة اﻻﺗﺟﺎه اﻟﺻﺣﯾﺢ ﻟﻌﻼﻣﺗﻲ اﻟﻘطﺑﯾﺔ )‪ +‬و ‪.(-‬‬‫ﻻ ﺗﺳﺗﺧدم ﺳوى ﺑطﺎرﯾﺎت ﻣﻘﺎس ‪ R6P‬أو ‪LR6‬‬‫أو ‪.AA‬‬‫اﻟﺧطوة ‪2‬‬‫‪ .3‬ر ّﻛب ﻏطﺎء اﻟﺑطﺎرﯾﺎت‪.‬‬‫ﺗﻧﺑﯾﮫ‬‫أدﺧل اﻟﺑطﺎرﯾﺎت اﻟﺟدﯾدة ﻛﻣﺎ ھو ﻣوﺿّﺢ ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت‪ .‬ﻟن ﯾﻌﻣل اﻟﺟﮭﺎز ﻓﻲ ﺣﺎﻟﺔ ﺗرﻛﯾﺑﮭﺎ ﺑطرﯾﻘﺔ‬‫ﻏﯾر ﺻﺣﯾﺣﺔ‪.‬‬‫)رﻣز‪ (LOW BATTERY‬ﻓﻲ اﻟﺷﺎﺷﺔ‪ ،‬اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺑطﺎرﯾﺎت ﺟدﯾدة‪ .‬ﻻ‬‫ﻋﻧدﻣﺎ ﺗُوﻣض‬‫ﺗﻌرض‬‫ﺗﺳﺗﺧدم ﺑطﺎرﯾﺎت ﺟدﯾدة ﻣﻊ أﺧرى ﻗدﯾﻣﺔ‪ .‬ﻗد ﯾﺗﺳﺑب ذﻟك ﻓﻲ ﺗﻘﺻﯾر ﻓﺗرة اﺳﺗﺧدام اﻟﺑطﺎرﯾﺎت‪ ،‬أو ﻓﻲ ّ‬‫اﻟﺟﮭﺎز ﻟﻠﻌطل‪.‬‬‫اﺳﺗﺑدل اﻟﺑطﺎرﯾﺎت ﺑﻌد إﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز ﺑﺛﺎﻧﯾﺗﯾن أو أﻛﺛر‪.‬‬‫)رﻣز ‪ (LOW BATTERY‬ﺣﺗﻰ ﺑﻌد اﺳﺗﺑدال اﻟﺑطﺎرﯾﺎت‪ ،‬ﻓﺄﺟ ِر ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫إذا ظﮭر‬‫ﯾﺗﻌرف اﻟﺟﮭﺎز ﺑﻌدﺋ ٍذ ﻋﻠﻰ اﻟﺑطﺎرﯾﺎت اﻟﺟدﯾدة‪.‬‬‫ﯾﻣﻛن أن‬‫ّ‬‫ﻻ ﯾظﮭر )رﻣز ‪ (LOW BATTERY‬ﻋﻧدﻣﺎ ﺗﻛون اﻟﺑطﺎرﯾﺎت ﻓﺎرﻏﺔ‪.‬‬‫ﺗﺗﺑﺎﯾن ﻓﺗرة ﺗﺷﻐﯾل اﻟﺑطﺎرﯾﺎت ﺣﺳب درﺟﺔ اﻟﺣرارة اﻟﻣﺣﯾطﺔ‪ ،‬وﻗد ﺗﻛون ﻓﺗرة ﺗﺷﻐﯾﻠﮭﺎ أﻗﺻر ﻋﻧد درﺟﺎت‬‫اﻟﺣرارة اﻟﻣﻧﺧﻔﺿﺔ‪ .‬وﺑﺷﻛل ﻋﺎم‪ ،‬ﺳﺗدوم اﻟﺑطﺎرﯾﺎت اﻷرﺑﻊ اﻟﺟدﯾدة ﻣﻘﺎس ‪ LR6‬ﻣدة ﻋﺎم واﺣد ﺗﻘرﯾﺑًﺎ ﻋﻧد‬‫اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻠﻘﯾﺎس ﻣرﺗﯾن ﻛل ﯾوم‪.‬‬‫ﻻ ﺗﺳﺗﺧدم ﺳوى اﻟﺑطﺎرﯾﺎت اﻟﻣﺣددة ﻓﻘط‪ .‬وﺗﻛون اﻟﺑطﺎرﯾﺎت اﻟﻣرﻓﻘﺔ ﻣﻊ اﻟﺟﮭﺎز ﻷﻏراض اﺧﺗﺑﺎر أداء ﺟﮭﺎز‬‫اﻟﻘﯾﺎس‪ ،‬وﯾﻣﻛن أن ﺗﻛون ذات ﻓﺗرة ﺗﺷﻐﯾل ﻣﺣدودة‪.‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت ﻓﻲ ﺣﺎﻟﺔ ﻋدم اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻔﺗرة طوﯾﻠﺔ‪.‬‬‫ﺗﺗﻌرض اﻟﺑطﺎرﯾﺎت ﻟﻠﺗﺳرﯾب ﻣﻣﺎ ﯾﺗﺳﺑب ﻓﻲ ّ‬‫ﺗﻌطل اﻟﺟﮭﺎز‪.‬‬‫ﯾﻣﻛن أن‬‫ّ‬‫اﻟﻌرﺑﯾﺔ ‪9‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﺗوﺻﯾل ﺧرطوم اﻟﮭواء‬‫ﻣﻘﺑس اﻟﮭواء‬‫ﺻل اﻟﮭواء ﻓﻲ ﻣﻘﺑس اﻟﮭواء‬‫أدﺧل ﻗﺎﺑس ﻣو ّ‬‫ﺑﺈﺣﻛﺎم‪.‬‬‫ﺻل اﻟﮭواء‬‫ﻗﺎﺑس ﻣو ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﺗوﺻﯾل ﻗﺎﺑس‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻓﻲ ﻣﻘﺑس اﻟﺗﯾﺎر‬‫أدﺧل ﻗﺎﺑس‬‫ّ‬‫اﻟﻣﺳﺗﻣر‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﺑﻣﺧرج اﻟﺗﯾﺎر‬‫ﺻل‬‫ﺑﻌد ذﻟك‪ِ ،‬‬‫ّ‬‫اﻟﻛﮭرﺑﺎﺋﻲ‪.‬‬‫ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫ﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻗﺎﺑس ﻣ ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد اﻟﻣﺣدد‪.‬‬‫اﺳﺗﺧدم ّ‬‫)راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪.(22‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻣن ﻣﺧرج اﻟﺗﯾﺎر اﻟﻛﮭرﺑﺎﺋﻲ‪ ،‬أﻣﺳك ھﯾﻛل‬‫ﻋﻧد ﻓﺻل‬‫ّ‬‫ّ‬‫واﺳﺣﺑﮫ ﺧﺎرج اﻟﻣﺧرج‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻣن ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬أﻣﺳك ﺑﻘﺎﺑس‬‫ﻋﻧد ﻓﺻل ﻗﺎﺑس‬‫ّ‬‫ّ‬‫اﻟﻣﺗردد واﺳﺣﺑﮫ ﺧﺎرج اﻟﺟﮭﺎز‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪10‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﺧﺗﯾﺎر اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟﻠرﺑﺎط‬‫إن اﺳﺗﻌﻣﺎل اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟرﺑﺎط اﻟذراع ھﺎم ﺟدًا ﻓﻲ دﻗّﺔ ﻗﯾﺎس ﺿﻐط اﻟدم‪ .‬وإذا ﻟم ﯾﻛن ﻣﻘﺎس اﻟرﺑﺎط‬‫ﺻﺣﯾ ًﺣﺎ‪ ،‬ﻓﻘﯾﺎس ﺿﻐط اﻟدم ﺳﯾظﮭر ﻗﯾﻣﺔ ﻏﯾر ﺻﺣﯾﺣﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﻋﺎ ﻋﻠﻰ ﻛل رﺑﺎط‪.‬‬‫ﯾﻛون ﻣﻘﺎس ﻣﺣﯾط اﻟذراع ﻣطﺑو ً‬‫ﯾﺳﺎﻋد ﻛ ﱞل ﻣن اﻟﻣؤﺷر واﻟﻧطﺎق اﻟﻣﻧﺎﺳب‪ ،‬اﻟﻣطﺑوﻋﺎن ﻋﻠﻰ اﻟرﺑﺎط‪ ،‬ﻋﻠﻰ أﻧك ﺗﺳﺗﺧدم اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ‪.‬‬‫)راﺟﻊ "اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ اﻟرﺑﺎط" ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ(‪.‬‬‫إذا ﻛﺎن اﻟﻣؤﺷر ﯾﺷﯾر إﻟﻰ ﺧﺎرج اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل اﻟﻣﺣﻠﻲ ﻟﺷراء رﺑﺎط ﺑدﯾل‪.‬‬‫رﺑﺎط اﻟذراع ھو أﺣد اﻟﻣواد اﻟﻣﺳﺗﮭﻠﻛﺔ‪ .‬ﻓﻲ ﺣﺎل أﺻﺑﺢ اﻟرﺑﺎط ﺑﺎﻟﯾﺎ ً ﻓﺎﺳْﺗﺑدﻟﮫ ﻓورا ً ﺑﺂﺧر ﺟدﯾد‪.‬‬‫ﺣﺟم اﻟرﺑﺎط اﻟﻣوﺻﻰ ﺑﮫ‬‫ﺣﺟم اﻟذراع‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬‫‪ 31‬ﺳم إﻟﻰ ‪ 45‬ﺳم‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬‫‪ 22‬ﺳم إﻟﻰ ‪ 42‬ﺳم‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪ 22‬ﺳم إﻟﻰ ‪ 32‬ﺳم‬‫ﺣﺟم اﻟذراع‪ :‬اﻟﻣﺣﯾط ﻋﻧد ﻋﺿﻠﺔ اﻟذراع‪.‬‬‫‪ UA-767S‬ﻏﯾر ﻣﺻﻣم ﻻﺳﺗﺧدام اﻟرﺑﺎط اﻟﺻﻐﯾر‪.‬‬‫ﻣﻠﺣوظﺔ‪:‬‬‫ﻛﯾﻔﯾﺔ وﺿﻊ رﺑﺎط اﻟذراع‬‫‪ .1‬ﻟف اﻟرﺑﺎط ﺣول أﻋﻠﻰ اﻟذراع )اﻟﻌﺿد(‪ ،‬ﺣواﻟﻲ‬‫‪ 2-1‬ﺳم أﻋﻠﻰ اﻟﺟزء اﻟداﺧﻠﻲ ﻣن اﻟﻣرﻓﻖ‪ ،‬ﻛﻣﺎ ھو‬‫ﻣوﺿﺢ‪.‬‬‫ﺿﻊ اﻟرﺑﺎط ﻣﺑﺎﺷرة ﻋﻠﻰ اﻟﺟﻠد‪ ،‬ﻷن اﻟﻣﻼﺑس ﯾﻣﻛﻧﮭﺎ‬‫أن ﺗﺗﺳﺑب ﻓﻲ اﻟﺣﺻول ﻋﻠﻰ ﻧﺑض ﺿﻌﯾف‪ ،‬وﯾﻧﺗﺞ‬‫ﻋﻧﮫ ﺧطﺄ ﻓﻲ اﻟﻘﯾﺎس‪.‬‬‫ﺿد‪ ،‬ﺑﺳﺑب‬‫‪ .2‬ﯾﻣﻛن أن ﯾﻣﻧﻊ اﻟﺗﺿﯾﯾﻖ اﻟﺷدﯾد ﻋﻠﻰ اﻟﻌ ُ‬‫ﻲ ﻛم اﻟﻘﻣﯾص ﺑﺷدة‪ ،‬اﻟﺣﺻول ﻋﻠﻰ ﻗراءات‬‫ط ّ‬‫دﻗﯾﻘﺔ‪.‬‬‫‪ .3‬ﺗﺄﻛد ﻣن وﺟود ﻧﻘﺎط اﻟﻣؤﺷر‬‫اﻟﻣﻧﺎﺳب‪.‬‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫ﻲ ﻛم اﻟﻘﻣﯾص ﺑﺷدة‪.‬‬‫ﻻ ﺗﻘم ﺑط ّ‬‫‪ 2-1‬ﺳم‬‫اﻟرﺑﺎط‬‫ﺧرطوم اﻟﮭواء‬‫رﻣز ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫ﺗﺄﻣﯾن اﻟﻧﺳﯾﺞ‬‫اﻟرﺑط ﺑﺈﺣﻛﺎم‪.‬‬‫ﺿﻣن اﻟﻧطﺎق‬‫اﻟﻣؤﺷر‬‫اﻟﻌرﺑﯾﺔ ‪11‬‬‫واﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ اﻟرﺑﺎط‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﺿﺑط اﻟﻌﻼﻣﺔ ﺑﺣﯾث ﺗﻛون ﻋﻧد‬‫ﺿد أو ﻓﻲ ﻣﺣﺎذاة ﻣﻊ‬‫ﺷرﯾﺎن اﻟﻌ ُ‬‫إﺻﺑﻊ اﻟﺑﻧﺻر ﻋﻧد اﻟﺟزء اﻟداﺧﻠﻲ ﻣن‬‫اﻟذراع‪.‬‬‫رﻣز ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫اﻟﻣؤﺷر‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟرﺑﺎط اﻟﺑﺎﻟﻐﯾن‪.‬‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن‪.‬‬‫ﯾﻛون ﻣطﺑو ً‬‫▲‬‫‪REF‬‬‫اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر ﺑدﻻً ﻣن‬‫ﺗﺟﺎوز اﻟﻧطﺎق اﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن واﻟرﺑﺎط‬‫ُ‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن أو اﻟرﺑﺎط ذي اﻟﻧطﺎق‬‫ذي اﻟﻧطﺎق اﻟﻌرﯾض‪.‬‬‫اﻟﻌرﯾض‪.‬‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟﻠرﺑﺎط ذي اﻟﻧطﻖ اﻟﻌرﯾض‪.‬‬‫ﻋﺎ ﻋﻠﻰ اﻟرﺑﺎط ذي اﻟﻧطﻖ اﻟﻌرﯾض‪.‬‬‫ﯾﻛون ﻣطﺑو ً‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟرﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪ .‬ﯾﻛون‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫ﻣطﺑو ً‬‫ﺿﻣن اﻟﻧطﺎق اﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن واﻟرﺑﺎط‬‫ذي اﻟﻧطﺎق اﻟﻌرﯾض‪.‬‬‫اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن ﺑدﻻً ﻣن رﺑﺎط‬‫ﺿﻣن اﻟﻧطﺎق اﻟﻣوﺟود ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫رﻗم اﻟﺗﺷﻐﯾﻠﺔ‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬

‫‪L‬‬

‫‪A‬‬
‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬

‫‪L‬‬

‫‪W‬‬

‫‪L‬‬

‫‪A‬‬

‫اﻟﻌرﺑﯾﺔ ‪12‬‬‫‪S‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪S‬‬

‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﻛﯾﻔﯾﺔ إﺟراء ﻋﻣﻠﯾﺎت ﻗﯾﺎس دﻗﯾﻘﺔ‬‫ﻟﻠﺣﺻول ﻋﻠﻰ ﻗﯾﺎس دﻗﯾﻖ ﺗﻣﺎ ًﻣﺎ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪:‬‬‫أرخ ذراﻋك ﻋﻠﻰ اﻟطﺎوﻟﺔ‪ .‬ﻻ ﺗﺿﻊ ﻗد ًﻣﺎ ﻓوق اﻷﺧرى‪ .‬أﺑﻖ ﻗدﻣﯾك ﻣﺳﺗوﯾﺗﯾن‬‫اﺟﻠس ﻣﺳﺗرﯾ ًﺣﺎ ﻋﻠﻰ ﻛرﺳﻲ‪.‬‬‫ِ‬‫ﻋﻠﻰ اﻷرض واﺟﻌل ظﮭرك ﻣﺳﺗﻘﯾ ًﻣﺎ‪.‬‬‫اﺳﺗرخ ﻟﻣدة ﺗﺗراوح ﺑﯾن ﺧﻣس إﻟﻰ ﻋﺷر دﻗﺎﺋﻖ ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫ِ‬‫اﺟﻌل ﻣﻧﺗﺻف اﻟرﺑﺎط ﻋﻧد اﻟﻣﺳﺗوى ﻧﻔﺳﮫ ﻟﻠﻘﻠب‪.‬‬‫اﺑﻖ ﺛﺎﺑﺗًﺎ وﺣﺎﻓظ ﻋﻠﻰ ھدوءك أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ﺳﺎ ﻣﺑﺎﺷرة ﺑﻌد ﻧﺷﺎط ﺑدﻧﻲ أو ﺑﻌد اﻻﻏﺗﺳﺎل‪ .‬اﺳﺗرخ ﻟﻣدة ﺗﺗراوح ﺑﯾن ﻋﺷرﯾن وﺛﻼﺛﯾن دﻗﯾﻘﺔ ﻗﺑل‬‫ﻻ ﺗﺟر ﻗﯾﺎ ً‬‫إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫ﺣﺎول ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ ﻛل ﯾوم‪.‬‬‫اﻟﻘﯾﺎس‬‫أﺛﻧﺎء اﻟﻘﯾﺎس‪ ،‬ﻣن اﻟطﺑﯾﻌﻲ أن ﺗﺷﻌر أن اﻟرﺑﺎط ﻣﺷدود ﻟﻠﻐﺎﯾﺔ‪) .‬ﻻ ﺗﺟزع(‪.‬‬‫ﺑﻌد اﻟﻘﯾﺎس‬‫ﺑﻌد اﻟﻘﯾﺎس‪ ،‬اﺿﻐط ﻋﻠﻰ اﻟزر ‪ START‬ﻹﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫اﻧزع اﻟرﺑﺎط وﺳﺟل ﺑﯾﺎﻧﺎﺗك‪.‬‬‫ﻲ دﻗﯾﻘﺔ واﺣد ﺗﻘرﯾﺑًﺎ ﻣن‬‫ﻣﻼﺣظﺔ‪ :‬ﯾﺗﻣﯾّز اﻟﺟﮭﺎز ﺑوظﯾﻔﺔ إﯾﻘﺎف اﻟﺗﺷﻐﯾل اﻟﺗﻠﻘﺎﺋﻲ‪ ،‬ﺣﯾث ﯾﻧطﻔﺊ اﻟﺟﮭﺎز ﺑﻌد ُﻣﺿ ّ‬‫اﻻﻧﺗﮭﺎء ﻣن ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫اﻧﺗظر ﻋﻠﻰ اﻷﻗل ﺛﻼث دﻗﺎﺋﻖ ﻗﺑل إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى ﻋﻠﻰ ﻧﻔس اﻟﺷﺧص‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪13‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‬‫ﻗﺑل اﻟﻘﯾﺎس‪ ،‬راﺟﻊ "ﻣﻼﺣظﺎت ﻟﻘﯾﺎس دﻗﯾﻖ" ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫اﻟﻘﯾﺎس اﻟﻌﺎدي‬‫‪.1‬‬‫ﺿﻊ اﻟرﺑﺎط ﻋﻠﻰ اﻟذراع )ﯾُﻔﺿل اﻟذراع اﻟﯾﺳرى(‪.‬‬‫اﺟﻠس ھﺎدﺋًﺎ ﻓﻲ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ زر ‪. START‬‬‫ﯾﺗم ﻋرض ﺟﻣﯾﻊ ﺷراﺋﺢ اﻟﺷﺎﺷﺔ‪.‬‬‫ﯾوﻣض اﻟﺻﻔر )‪ (0‬ﻋﻠﻰ اﻟﺷﺎﺷﺔ ﻟﻔﺗرة وﺟﯾزة‪.‬‬‫ﺗﺗﻐﯾّر اﻟﺷﺎﺷﺔ‪ ،‬ﻛﻣﺎ ھو ﻣوﺿّﺢ ﻓﻲ اﻟﺷﻛل ﻋﻠﻰ اﻟﯾﻣﯾن‪ ،‬ﺣﯾث‬‫ﺗﺑدأ ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬ﯾﺑدأ اﻟرﺑﺎط ﻓﻲ اﻻﻧﺗﻔﺎخ‪ .‬ﻣن اﻟطﺑﯾﻌﻲ أن‬‫ﺗﺷﻌر أن اﻟرﺑﺎط ﻣﺷدود ﻟﻠﻐﺎﯾﺔ‪ .‬ﯾُﻌرض ﻣؤﺷر ﺷرﯾط‬‫اﻟﺿﻐط‪ ،‬ﻋﻠﻰ اﻟﺣﺎﻓﺔ اﻟﯾﺳرى ﻣن اﻟﺷﺎﺷﺔ‪ ،‬ﻓﻲ أﺛﻧﺎء ﻧﻔﺦ‬‫اﻟﮭواء‪.‬‬‫ﻣﻠﺣوظﺔ‪ :‬إذا أردت إﯾﻘﺎف ﻧﻔﺦ اﻟﮭواء ﻓﻲ أي وﻗت‪ ،‬ﻓﺎﺿﻐط‬‫ﻋﻠﻰ زر ‪ START‬ﻣرة أﺧرى‪.‬‬‫‪.3‬‬‫ﻋﻧد اﻛﺗﻣﺎل ﻧﻔﺦ اﻟﮭواء‪ ،‬ﯾﺑدأ اﻻﻧﻛﻣﺎش أوﺗوﻣﺎﺗﯾﻛﯾًﺎ وﯾوﻣض‬‫ﺟﺎر إﺟراء اﻟﻘﯾﺎس‪ .‬ﺑﻣﺟرد‬‫)رﻣز اﻟﻘﻠب(‪،‬‬‫ً‬‫ﻣﺷﯾرا إﻟﻰ أﻧﮫ ٍ‬‫ﺗﺣدﯾد اﻟﻧﺑض‪ ،‬ﯾوﻣض اﻟرﻣز ﻣﻊ ﻛل ﻧﺑﺿﺔ‪.‬‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫‪START‬‬‫اﺿﻐط‬‫ﻛل اﻟﺷﺎﺷﺔ‬‫اﻷﺟزاء اﻟﻣﻌروﺿﺔ‬‫ﻋرض اﻟﺻﻔر‬‫ﺑدء ﻧ ْﻔﺦ اﻟﮭواء‬‫ﺟﺎر ﻗﯾﺎس اﻟﺿﻐط‬‫ٍ‬‫ﻣﻼﺣظﺔ‪ :‬ﻓﻲ ﺣﺎﻟﺔ ﻋدم اﻟﺣﺻول ﻋﻠﻰ ﺿﻐط ﻣﻼﺋم‪ ،‬ﯾﺑدأ‬‫اﻟﺟﮭﺎز ﻓﻲ ﻧ ْﻔﺦ اﻟﮭواء ﻣرة أﺧرى ﺗﻠﻘﺎﺋﯾًﺎ‪.‬‬‫ﻟﺗﺟﻧب إﻋﺎدة ﻧﻔﺦ اﻟﮭواء‪ ،‬راﺟﻊ "إﺟراء اﻟﻘﯾﺎس ﻣﻊ‬‫اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ" ﻓﻲ‬‫اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫‪.4‬‬‫ﻋﻧد اﻻﻧﺗﮭﺎء ﻣن اﻟﻘﯾﺎس‪ ،‬ﺗﻌرض اﻟﺷﺎﺷﺔ اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪،‬‬‫واﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ ،‬وﺳرﻋﺔ اﻟﻧﺑض‪.‬‬‫ﯾﻔرغ اﻟرﺑﺎط اﻟﮭواء اﻟﻣﺗﺑﻘﻲ ﺗﻣﺎ ًﻣﺎ‪.‬‬‫ّ‬‫‪.5‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر ‪ START‬ﻹﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫إﺟراء اﻟﻘﯾﺎس ﻗﯾد‬‫اﻟﺗﻧﻔﯾذ‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﻣﻼﺣظﺔ‪ :‬اﺟﻌل اﻟﻣدة اﻟزﻣﻧﯾﺔ اﻟﻔﺎرﻗﺔ ﺑﯾن ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻧﻔس‬‫اﻟﺷﺧص ﺛﻼث دﻗﺎﺋﻖ ﻋﻠﻰ اﻷﻗل‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪14‬‬‫ﺳرﻋﺔ اﻟﻧﺑض‬‫ﺗﻔرﯾﻎ اﻟﮭواء اﻟﻣﺗﺑﻘﻲ‬‫ﺗﻠﻘﺎﺋﯾًﺎ‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‬‫إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ‬‫‪ UA-767S‬ﻣﺻﻣم ﻟﻠﺗﻌرف ﻋﻠﻰ اﻟﻧﺑض وﻟﻧﻔﺦ اﻟرﺑﺎط ﺑﺎﻟﮭواء إﻟﻰ ﻣﺳﺗوى اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫اﺳﺗﺧدم ھذه اﻟطرﯾﻘﺔ ﻋﻧد ﺗﻛرار ﻧﻔﺦ اﻟﮭواء أو ﻋﻧد ﻋدم ﻋرض اﻟﻧﺗﺎﺋﺞ ﺣﺗﻰ ﻋﻧد اﻧﺧﻔﺎض ﻣﺳﺗوى اﻟﺿﻐط‬‫إﻟﻰ ‪ mmHg 20‬أو أﻗل‪.‬‬‫‪.1‬‬‫وﺿﻊ اﻟرﺑﺎط ﻋﻠﻰ اﻟذراع ﻓﻲ ﻣﺳﺗوى اﻟﻘﻠب )ﯾﻔﺿل‬‫اﻟذراع اﻟﯾﺳرى(‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ زر‬‫‪.3‬‬‫ﻋﻧدﻣﺎ ﯾوﻣض اﻟرﻗم ﺻﻔر‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ‬‫زر ‪ START‬ﺣﺗﻰ ﯾظﮭر رﻗم أﻋﻠﻰ ﻣن ﺿﻐطك‬‫اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣﺗوﻗﻊ ﺑـ ‪ 30‬إﻟﻰ ‪.mmHg 40‬‬‫ﺣرر اﻟزر ﻋﻧد ﻣﺳﺗوى‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﻣطﻠوب‬‫‪.4‬‬‫ﻋﻧد ﺑﻠوغ اﻟرﻗم اﻟﻣطﻠوب‪ ،‬ﺣرر اﻟزر ‪ START‬ﻟﺑدء‬‫اﻟﻘﯾﺎس‪ .‬ﺗﺎﺑﻊ إﺟراء ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻛﻣﺎ‬‫ھو ﻣوﺿﺢ ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺳﺎﺑﻘﺔ‪.‬‬‫اﺿﻐط ﺑﺎﺳﺗﻣرار‬‫ﻋﻠﻰ اﻟزر‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫‪START‬‬‫‪. START‬‬‫‪START‬‬‫اﺿﻐط‬‫اﻟزر‬‫راﺟﻊ اﻟﺻﻔﺣﺔ اﻟﺳﺎﺑﻘﺔ ﻟﻼطﻼع ﻋﻠﻰ‬‫إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﻣﻼﺣظﺎت ﻟﻘﯾﺎس دﻗﯾﻖ‬‫اﺟﻠس ﻓﻲ وﺿﻊ ﻣرﯾﺢ‪ .‬ﺿﻊ ذراﻋك ﻋﻠﻰ طﺎوﻟﺔ ﺑﺣﯾث ﺗﻛون راﺣﺔ ﯾدك ﻣواﺟﮭﺔ ﻷﻋﻠﻰ واﻟرﺑﺎط ﻋﻧد‬‫ﻣﺳﺗوى اﻟﻘﻠب‪.‬‬‫اﺳﺗرخ ﻟﻣدة ﺣواﻟﻲ ﻣن ﺧﻣس إﻟﻰ ﻋﺷر دﻗﺎﺋﻖ ﻗﺑل إﺟراء اﻟﻘﯾﺎس‪ .‬إذا ﻛﺎﻧت ﺗﻌﺗرﯾك ﻓرﺣﺔ ﻋﺎرﻣﺔ أو ﺗﻣر‬‫ﺑﺣﺎﻟﺔ اﻛﺗﺋﺎب ﻧﺗﯾﺟﺔ ﻟﻠﺿﻐوطﺎت اﻟﻌﺎطﻔﯾﺔ‪ ،‬ﻓﺳﯾﻧﻌﻛس ذﻟك ﻣﻧﺗ ًﺟﺎ ﻗراءة أﻋﻠﻰ )أو أﻗل( ﻣن اﻟﻘراءة اﻟطﺑﯾﻌﯾﺔ‬‫ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬وﺳﺗﻛون ﻗراءة ﺳرﻋﺔ اﻟﻧﺑض أﺳرع ﻣن اﻟﻣﻌدل اﻟطﺑﯾﻌﻲ‪.‬‬‫ﯾﺗﺑﺎﯾن ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟﻠﻔرد ﺑﺎﺳﺗﻣرار ﺗﺑﻌًﺎ ﻟﻠﻌﻣل اﻟذي ﺗﻘوم ﺑﮫ واﻟطﻌﺎم اﻟذي ﺗﺗﻧﺎوﻟﮫ‪ .‬ﯾﻣﻛن أن ﯾﻛون ﻟﻣﺎ‬‫ﺗﺷرﺑﮫ ﺗﺄﺛﯾر ﻗوي وﺳرﯾﻊ ﻋﻠﻰ ﺿﻐط دﻣك‪.‬‬‫ﯾﻌﺗﻣد ھذا اﻟﺟﮭﺎز ﻓﻲ ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس اﻟﺗﻲ ﯾُﺟرﯾﮭﺎ ﻋﻠﻰ ﺿرﺑﺎت اﻟﻘﻠب‪ .‬ﻓﺈذا ﻛﺎن ﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟدﯾك‬‫ﺿﻌﯾﻔًﺎ ﺟدًا أو ﻏﯾر ﻣﻧﺗظم‪ ،‬ﻓﻘد ﯾﺟد اﻟﺟﮭﺎز ﺻﻌوﺑﺔ ﻓﻲ ﺗﺣدﯾد ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪.‬‬‫إذا اﻛﺗﺷف اﻟﺟﮭﺎز ﺣﺎﻟﺔ ﻏﯾر طﺑﯾﻌﯾﺔ‪ ،‬ﻓﺳﯾوﻗف ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس وﯾُظﮭر رﻣز اﻟﺧطﺄ‪ .‬راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪7‬‬‫ﻟوﺻف اﻟرﻣوز‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪15‬‬‫ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ُﻣﺧﺻص ﻟﻼﺳﺗﺧدام ﻣن ﻗِﺑل اﻟﺑﺎﻟﻐﯾن‪ .‬اﺳﺗﺷر طﺑﯾﺑك ﻗﺑل اﺳﺗﺧدام ھذا‬‫اﻟﺟﮭﺎز ﻣﻊ أﺣد اﻷطﻔﺎل‪ .‬ﯾﺟب ﻋدم اﺳﺗﺧدام اﻷطﻔﺎل اﻟﺟﮭﺎز دون إﺷراف‪.‬‬‫ﯾﻣﻛن أن ﯾﺗﺄﺛر أداء ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﺗﻠﻘﺎﺋﻲ ﺑﺳﺑب اﻟزﯾﺎدة اﻟﺑﺎﻟﻐﺔ ﻓﻲ درﺟﺔ اﻟﺣرارة أو‬‫اﻟرطوﺑﺔ أو اﻻرﺗﻔﺎع ﻋن ﺳطﺢ اﻟﺑﺣر‪.‬‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة‬‫ﻣﻼﺣظﺔ‪ :‬ﯾﺧزن ھذا اﻟﺟﮭﺎز آﺧر ‪ 60‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻓﻲ اﻟذاﻛرة‪.‬‬‫اﻟزر‬‫‪ .1‬ﻋﻧد ﻋدم ظﮭور أي ﺷﻲء‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار‬‫ﻋﻠﻰ زر ‪ START‬ﻻﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت‬‫اﻟ ُﻣﺧزﻧﺔ‪.‬‬‫‪START‬‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﺿﻐط‬‫ﺑﺎﺳﺗﻣرار‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﻧﺑض‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫ﻟﻛل اﻟﺑﯾﺎﻧﺎت‬‫ﺣرر اﻟزر ﻋﻧد ظﮭور ﻣﺗوﺳط اﻟﻘﯾﺎﺳﺎت‬‫‪.2‬‬‫ّ‬‫اﻟﺳﺎﺑﻘﺔ اﻟﻣﺧزﻧﺔ‪.‬‬‫‪ .3‬ﺗُﻌرض رﻗم اﻟﻘﯾﺎﺳﺎت واﻟﻘﯾﺎﺳﺎت اﻟ ُﻣﺧزﻧﺔ ﺗﻠﻘﺎﺋﯾًﺎ‬‫ﺑﺎﻟﺗرﺗﯾب وذﻟك ﻣن آﺧر ﻣرة ﻗﯾﺎس‪.‬‬‫اﻧﻘﺑﺎﺿﻲ‬‫اﻧﺑﺳﺎطﻲ‬‫‪ .4‬ﺳﺗﻧطﻔﺊ اﻟﺷﺎﺷﺔ أوﺗوﻣﺎﺗﯾﻛﯾًﺎ ﺑﻌد ﻋرض ﻛل اﻟﺑﯾﺎﻧﺎت‪.‬‬‫اﻟﻧﺑض‬‫أﺣدث اﻟﺑﯾﺎﻧﺎت‬‫ﻣﻠﺣوظﺔ‪ :‬إذا ﺿﻐطت ﻋﻠﻰ زر ‪ START‬ﻓﻲ أﺛﻧﺎء‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت‪ ،‬ﻓﺳﯾﻧطﻔﺊ اﻟﺟﮭﺎز‪.‬‬‫اﻧﻘﺑﺎﺿﻲ‬‫اﻧﺑﺳﺎطﻲ‬‫اﻟﻧﺑض‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻘدﯾﻣﺔ )اﻷﺧﯾرة(‬‫اﻟﻌرﺑﯾﺔ ‪16‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣؤﺷر ‪IHB/AFib‬؟‬‫ﻋﻧد اﻛﺗﺷﺎف ﺟﮭﺎز اﻟﻘﯾﺎس وﺟود ْ‬‫ﻧظم ﻏﯾر ﻣﻧﺗظم أﺛﻧﺎء ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‪ ،‬ﺳﯾظﮭر اﻟﻣؤﺷر ‪ IHB/AFib‬ﻋﻠﻰ‬‫اﻟﺷﺎﺷﺔ ﻣﻊ ﻗﯾم اﻟﻘﯾﺎس‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬ﻧﺣن ﻧوﺻﻲ ﺑﺎﻻﺗﺻﺎل ﺑﺎﻟطﺑﯾب إذا ظﮭر ﻟك ﻣؤﺷر‬‫‪ IHB/AFib‬ھذا ﺑﺻورة ﻣﺗﻛررة‪.‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑـ ‪AFib‬؟‬‫ظرا إﻟﻰ اﻹﺷﺎرات اﻟﻛﮭرﺑﺎﺋﯾﺔ اﻟواﻗﻌﺔ ﻓﻲ اﻟﻘﻠب وﯾﻧﺷر اﻟدم ﻓﻲ ﺟﻣﯾﻊ أﻧﺣﺎء اﻟﺟﺳم‪ .‬ﯾﺣدث‬‫ﯾﻧﻘﺑض اﻟﻘﻠب ﻧ ً‬‫اﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪ (AFib‬ﻋﻧدﻣﺎ ﺗُﺻﺑﺢ اﻹﺷﺎرة اﻟﻛﮭرﺑﺎﺋﯾﺔ ﻓﻲ اﻷُذﯾن ﻣﺷوﺷﺔ‪ ،‬ﻣﺎ ﯾؤدي إﻟﻰ اﺿطراﺑﺎت‬‫ﻓﻲ اﻟﻔواﺻل اﻟزﻣﻧﯾﺔ ﺑﯾن اﻟﺿرﺑﺎت‪ .‬وﯾﻣﻛن أن ﯾﺳﺑب اﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪ (AFib‬رﻛود اﻟدم ﻓﻲ اﻟﻘﻠب‪ ،‬ﻣﺎ‬‫اﻟﺗﻌرض ﻟﻧوﺑﺔ ﻗﻠﺑﯾﺔ‪.‬‬‫ﯾﺷ ّﻛل ﺑﺳﮭوﻟﺔ ﺟﻠطﺎت دﻣوﯾﺔ ﺗﺗﺳﺑب ﻻﺣﻘًﺎ ﻓﻲ‬‫ّ‬‫‪％IHB/AFib‬‬‫ﺗُﻌرض ‪ %IHB/AFib‬ﻛﺗﻛرار ﻟـ ‪ IHB/AFib‬اﻟﺗﻲ ﺗم رﺻدھﺎ‪.‬‬‫ﯾﻣﻛن ﻟـ ‪ IHB/AFib‬رﺻد أﺷﯾﺎء أﺧرى ﻏﯾر اﻟﺿوﺿﺎء‪ ،‬ﻣﺛل اﻟﺣرﻛﺔ اﻟﺟﺳدﯾﺔ وﻋدم اﻧﺗظﺎم ﺿرﺑﺎت‬‫اﻟﻘﻠب‪ .‬ﻟذﻟك‪ ،‬ﻧوﺻﻲ ﺑﺎﻟﺗواﺻل ﻣﻊ طﺑﯾﺑك إذا ﻛﺎن ﻣﺳﺗوى ‪ %IHB/AFib‬ﻣرﺗﻔﻌًﺎ‪.‬‬‫ﻋدد ﺣﺎﻻت ‪IHB/AFib‬‬‫اﻟ ُﻣﻛﺗﺷﻔﺔ ﻓﻲ اﻟذاﻛرة‬‫‪= %IHB/AFib‬‬‫‪[%] 100 x‬‬‫اﻟرﻗم اﻟﻛﻠﻲ‬‫ﻋرض ‪ :%IHB/AFib‬ﺗُﻌرض ‪ %IHB/AFib‬ﻋﻧد ﻋرض اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﻘﯾم‪.‬‬‫)راﺟﻊ اﻟﺻﻔﺣﺔ "رﻗم ‪ 2‬اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت" ﻓﻲ"ﻧﻣط اﻟﺗﺷﻐﯾل"(‬‫ﻻ ﯾﺗم ﻋرض ‪ %IHB/AFib‬ﻋﻧدﻣﺎ ﯾﻛون رﻗم اﻟذاﻛرة ﺳﺗﺔ أو أﻗل‪.‬‬‫ﻋرض اﻟﻘﯾم اﻟﻣﺗوﺳطﺔ‬‫اﻟﻣﺳﺗوى ‪0‬‬‫‪0=%IHB/AFib‬‬‫اﻟﻣﺳﺗوى ‪1‬‬‫‪9 - 1=%IHB/AFib‬‬‫اﻟﻣﺳﺗوى ‪2‬‬‫‪24 - 10=%IHB/AFib‬‬‫ﻻ ﯾُﻌرض ﺷﻲء‬‫اﻟﻌرﺑﯾﺔ ‪17‬‬‫اﻟﻣﺳﺗوى ‪3‬‬‫‪100 - 25=%IHB/AFib‬‬

‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‬‫ﯾراﻗب اﻟﻣؤﺷر ﻣدى اﻟﺗﻘدّم ﻓﻲ ﻋﻣﻠﯾﺔ ﻗﯾﺎس اﻟﺿﻐط‪.‬‬‫ﺟﺎر ﻗﯾﺎس اﻟﺿﻐط‬‫ٍ‬‫ﺟﺎر ﺗﻔرﯾﻎ اﻟﮭواء‬‫ٍ‬‫ﻧ ْﻔﺦ اﻟﮭواء ﻗﯾد اﻟﺗﻧﻔﯾذ‬‫ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‬‫اﻛﺗﻣل ﻧ ْﻔﺦ اﻟﮭواء‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﯾﺷﯾر ﻛل ﺟزء ﻓﻲ ﺷرﯾط اﻟﻣؤﺷر إﻟﻰ ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ‬‫اﻟﻌﺎﻟﻣﯾﺔ ﻟﺿﻐط اﻟدم اﻟﻣوﺻوف ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﺷدﯾد‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﻣﺗوﺳط‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﺧﻔﯾف‬‫ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫طﺑﯾﻌﻲ‬‫ﻣﺛﺎﻟﻲ‬‫‪ :‬ﯾﻌرض اﻟﻣؤﺷر ﺷرﯾﺣﺔ ﺗﺑﻌًﺎ ﻟﻠﺑﯾﺎﻧﺎت اﻟﺣﺎﻟﯾﺔ وﺑﻣﺎ ﯾﺗواﻓﻖ ﻣﻊ‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‪.‬‬‫ﻣﺛﺎل‪:‬‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﻣﺗوﺳط‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﺧﻔﯾف‬‫ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫ﻧُﺑذة ﻋن ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣﺳﺗوى ﺿﻐط اﻟدم؟‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم ھو ﻗوة دﻓﻊ اﻟدم ﻟﺟدران اﻟﺷراﯾﯾن‪ .‬وﻋﻧدﻣﺎ ﯾﻧﻘﺑض اﻟدم‪ ،‬ﯾﺣدث اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪.‬‬‫وﻋﻧدﻣﺎ ﯾﻧﺑﺳط اﻟدم‪ ،‬ﯾﺣدث اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ .‬وﯾﻛون اﻟﻣﻠﻠﯾﻣﺗر اﻟزﺋﺑﻘﻲ )‪ (mmHg‬وﺣدة ﻗﯾﺎس ﻣﺳﺗوى‬‫ﺿﻐط اﻟدم‪ .‬وﯾُﻣﺛل اﻟﺿﻐط اﻷﺳﺎﺳﻲ ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟطﺑﯾﻌﻲ ﻟﻠﻔرد‪ ،‬وھو ّأول ﻣﺎ ﯾُﻘﺎس ﻓﻲ اﻟﺻﺑﺎح‬‫ﺣﯾث ﯾﻛون اﻟﻔرد ﻣﺳﺗرﺧﯾًﺎ وﻗﺑل ﺗﻧﺎوﻟﮫ اﻟطﻌﺎم‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪18‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎرﺗﻔﺎع ﺿﻐط اﻟدم وﻛﯾف ﯾﻣﻛن اﻟﺗﺣﻛم ﻓﯾﮫ؟‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم ھو ﺣﺎﻟﺔ ﻣن ﺿﻐط اﻟدم اﻟﺷرﯾﺎﻧﻲ اﻟﻣرﺗﻔﻊ ﻏﯾر اﻟطﺑﯾﻌﻲ‪ ،‬اﻟذي ﯾﻣﻛن أن ﯾﺗﺳﺑب‪ ،‬ﻓﻲ ﺣﺎﻟﺔ‬‫ﻟﺗﻌرض ﻟﻣﺷﻛﻼت ﺻﺣﯾﺔ‪ ،‬ﻣن ﺑﯾﻧﮭﺎ اﻟﻧوﺑﺔ اﻟﻘﻠﺑﯾﺔ‪ .‬وﯾﻣﻛن اﻟﺗﺣﻛم ﻓﻲ ارﺗﻔﺎع ﺿﻐط‬‫ﻋدم اﻟﺗﺣﻛم ﻓﯾﮫ‪ ،‬ﻓﻲ ا ّ‬
‫اﻟدم ﻋن طرﯾﻖ ﺗﻐﯾﯾر ﻧﻣط اﻟﺣﯾﺎة‪ ،‬وﺗﻔﺎدي ﺣﺎﻻت اﻹﺟﮭﺎد‪ ،‬وﺑﺗﻧﺎول اﻟﻌﻼج ﺗﺣت اﻹﺷراف اﻟطﺑﻲ‪.‬‬‫ﻟﻠوﻗﺎﯾﺔ ﻣن ارﺗﻔﺎع ﺿﻐط اﻟدم أو اﻟﺗﺣﻛم ﻓﯾﮫ‪:‬‬‫ﻣﺎرس اﻟرﯾﺎﺿﺔ ﺑﺎﻧﺗظﺎم‬‫ﺗﺟﻧب اﻟﺗدﺧﯾن‬‫اﺧﺿﻊ ﻟﻔﺣوﺻﺎت طﺑﯾﺔ ﻣﻧﺗظﻣﺔ‬‫ﻗﻠّل ﻣن ﺗﻧﺎول اﻷﻣﻼح واﻟدھون‬‫ﺣﺎﻓظ ﻋﻠﻰ وزﻧك اﻟﻣﺛﺎﻟﻲ‬‫ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟‬‫إن ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ ﻋﯾﺎدة أو ﻋﻧد اﻟطﺑﯾب ﯾﻣﻛن أن ﯾﺛﯾر اﻟﻣﺧﺎوف وﯾُﻌطﻲ ﻗراءة ﻣرﺗﻔﻌﺔ‪،‬‬‫ﺑﻣﻘدار ‪ 25‬إﻟﻰ ‪ mmHg 30‬أﻋﻠﻰ ﻣن اﻟﻘﯾﺎس ﻓﻲ اﻟﻣﻧزل‪ .‬ﯾﻘﻠل اﻟﻘﯾﺎس ﻓﻲ اﻟﻣﻧزل ﻣن اﻟﺗﺄﺛﯾرات اﻟﺧﺎرﺟﯾﺔ‬‫ﻋﻠﻰ ﻗراءات ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬وﯾﻣﺛل إﺿﺎﻓﺔ إﻟﻰ ﻗراءات اﻟطﺑﯾب‪ ،‬وﯾﻘدم ﺳﺟﻼً ﻣن اﻟﻘراءات اﻟﻛﺎﻣﻠﺔ‬‫واﻷﻛﺛر دﻗﺔ ﺑﺎﻟﻧﺳﺑﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‬‫وﺿﻌت ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻣﻌﺎﯾﯾر‬‫ﻟﺗﻘﯾﯾم ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ‪ ،‬دون‬‫اﻟﻧظر إﻟﻰ اﻟﻌﻣر‪ ،‬ﻛﻣﺎ ﯾﺗﺿﺢ ذﻟك ﻓﻲ‬‫اﻟرﺳم اﻟﺗﺧطﯾطﻲ‪.‬‬‫اﻻﺧﺗﻼﻓﺎت ﻓﻲ ﺿﻐط اﻟدم‬‫ﯾﺧﺗﻠف ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟﻠﻔرد اﺧﺗﻼﻓًﺎ‬‫ﻛﺑﯾرا ﻋﻠﻰ ﻣدار اﻟﯾوم واﻟﻣوﺳم‪ .‬وﯾﻣﻛن‬‫ً‬‫أن ﯾﺗراوح ﻣﻘدار اﻻﺧﺗﻼف ﺑﯾن‬‫‪ 30‬و‪ mmHg 50‬ﻧظ ًرا إﻟﻰ اﻟظروف‬‫اﻟﻣﺧﺗﻠﻔﺔ طوال اﻟﯾوم‪ .‬وﯾﻣﻛن أن ﺗﻛون‬‫اﻻﺧﺗﻼﻓﺎت ﻟدى اﻷﻓراد اﻟذﯾن ﯾﻌﺎﻧون ارﺗﻔﺎع ﺿﻐط اﻟدم ﻣﻠﺣوظﺔ ﺑﻘدر أﻛﺑر‪ .‬وﻋﺎدة ﻣﺎ ﯾرﺗﻔﻊ ﻣﺳﺗوى‬‫ﻛﺛﯾرا‬‫ﺿﻐط اﻟدم ﻓﻲ أﺛﻧﺎء اﻟﻌﻣل أو اﻟﻠﻌب‪ ،‬وﯾﻧﺧﻔض إﻟﻰ أدﻧﻰ ﻣﺳﺗوﯾﺎﺗﮫ ﻓﻲ أﺛﻧﺎء اﻟﻧوم‪ .‬ﻟذﻟك‪ ،‬ﻻ ﺗﻘﻠﻖ‬‫ً‬‫ﺣﯾﺎل اﻟﻧﺗﺎﺋﺞ اﻟﺗﻲ ﺗظﮭر ﻣن ﻋﻣﻠﯾﺔ ﻗﯾﺎس واﺣدة‪.‬‬‫أﺟر ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ ﻛل ﯾوم ﺑﺎﺗﺑﺎع‬‫ِ‬‫اﻹﺟراء اﻟﻣوﺿّﺢ ﻓﻲ ھذا اﻟدﻟﯾل ﻟﻣﻌرﻓﺔ اﻟﻣﻌدل‬‫اﻟطﺑﯾﻌﻲ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪ .‬وﺗﻘدّم اﻟﻘراءات‬‫اﻟﻣﻧﺗظﻣﺔ ﺳﺟﻼً أﻛﺛر ﺷﻣوﻟﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫اﺣرص ﻋﻠﻰ ﺗوﺿﯾﺢ اﻟوﻗت واﻟﺗﺎرﯾﺦ ﻋﻧد ﺗﺳﺟﯾل‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪ .‬اﺳﺗﺷر اﻟطﺑﯾب ﻟﯾﻔﺳّر ﻟك‬‫ﺑﯾﺎﻧﺎت ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪19‬‬‫اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ‬‫اﻟﻣﺷﻛﻠﺔ‬‫ﻻ ﯾظﮭر ﺷﻲء ﻋﻠﻰ‬‫اﻟﺷﺎﺷﺔ‪ ،‬ﺣﺗﻰ ﻋﻧد‬‫ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫اﻟﺳﺑب اﻟﻣﺣﺗﻣل‬‫ﻧﻔدت طﺎﻗﺔ اﻟﺑطﺎرﯾﺎت‪.‬‬‫أطراف اﻟﺑطﺎرﯾﺎت ﻟﯾﺳت ﻓﻲ‬‫ﻣواﺿﻌﮭﺎ اﻟﺻﺣﯾﺣﺔ‪.‬‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪.‬‬‫أ ِﻋد ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت ﺑﺣﯾث ﺗﻛون اﻷطراف‬‫اﻟﺳﺎﻟﺑﺔ واﻟﻣوﺟﺑﺔ ﻣﺗطﺎﺑﻘﺔ ﻣﻊ ﺗﻠك اﻟﻣوﺿﺣﺔ‬‫ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت‪.‬‬‫ﺟﮭد اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔض ﻟﻠﻐﺎﯾﺔ‪.‬‬‫)رﻣز ‪LOW‬‬‫ﯾوﻣض‬‫ﻻ ﯾﻧﺗﻔﺦ اﻟرﺑﺎط‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪.‬‬‫‪ .(BATTERY‬إذا ﻧﻔدت‬‫ﺑﺎﻟﮭواء‪.‬‬‫اﻟﺑطﺎرﯾﺎت ﺗﻣﺎ ًﻣﺎ‪ ،‬ﻓﻠن ﯾظﮭر اﻟرﻣز‪.‬‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﻟم ﯾُﻠف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﻣﻼﺋﻣﺔ‪.‬‬‫ﺣرﻛت ذراﻋك أو ﺟﺳدك ﻓﻲ أﺛﻧﺎء‬‫ّ‬‫اﺣرص ﻋﻠﻰ أن ﺗظل ﺛﺎﺑﺗًﺎ وھﺎدﺋًﺎ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﻟﻘﯾﺎس‪.‬‬‫اﺟﻠس ﻓﻲ ﺣﺎﻟﺔ ﺛﺑﺎت وھدوء‪.‬‬‫اﻟﺟﮭﺎز ﻻ ﯾﻘﯾس‪.‬‬‫ﺿﻊ ذراﻋك ﻋﻠﻰ طﺎوﻟﺔ ﺑﺣﯾث ﺗﻛون راﺣﺔ‬‫اﻟﻘراءات ﻣرﺗﻔﻌﺔ أو‬‫ﻣوﺿﻊ اﻟرﺑﺎط ﻏﯾر ﺻﺣﯾﺢ‪.‬‬‫ﯾدك ﻣواﺟﮭﺔ ﻷﻋﻠﻰ واﻟرﺑﺎط ﻋﻧد ﻣﺳﺗوى‬‫ﻣﻧﺧﻔﺿﺔ ﻟﻠﻐﺎﯾﺔ‪.‬‬‫اﻟﻘﻠب‪.‬‬‫إذا ﻛﺎن ﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟدﯾك ﺿﻌﯾﻔًﺎ ﺟدًا‬‫أو ﻏﯾر ﻣﻧﺗظم‪ ،‬ﻓﻘد ﯾﺟد اﻟﺟﮭﺎز ﺻﻌوﺑﺔ ﻓﻲ‬‫ﺗﺣدﯾد ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪.‬‬‫اﻟﻘﯾﻣﺔ ﻣﺧﺗﻠﻔﺔ ﻋن اﻟﻘﯾﺎس ﻓﻲ اﻟﻌﯾﺎدة‬‫راﺟﻊ "ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟"‪.‬‬‫أو ﻟدى اﻟطﺑﯾب‪.‬‬‫أﺧرى‬‫اﻧزع اﻟﺑطﺎرﯾﺎت‪ .‬ﺿﻌﮭﺎ ﻣرة أﺧرى ﺑﺷﻛل‬‫ﺻﺣﯾﺢ و ﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة‬‫أﺧرى‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬إذا ﻟم ﺗُﺣل اﻟﻣﺷﻛﻼت ﺑﺎﺗﺑﺎع اﻹﺟراءات اﻟﻣوﺿﺣﺔ أﻋﻼه‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل‪ .‬ﻻ ﺗﺣﺎول ﻓﺗﺢ أو إﺻﻼح‬‫ﻧظرا إﻟﻰ أن أي ﻣﺣﺎوﻟﺔ ﻟﻠﻘﯾﺎم ﺑذﻟك ﺳﺗؤدي إﻟﻰ إﻟﻐﺎء اﻟﺿﻣﺎن‪.‬‬‫ھذا اﻟﻣﻧﺗﺢ‪ً ،‬‬‫اﻟﻌرﺑﯾﺔ ‪20‬‬‫اﻟﺻﯾﺎﻧﺔ‬‫ﻣﻛوﻧﺎت ﻛﮭرﺑﺎﺋﯾﺔ ﺷدﯾدة اﻟﺣﺳﺎﺳﯾﺔ ووﺣدة ھواء ﻣﻌﻘدة ﯾﻣﻛن أن ﺗﺗﻠف‪ .‬وإذا ﺗﻌذّر‬‫ﻻ ﺗﻔﺗﺢ اﻟﺟﮭﺎز‪ .‬ﺗُﺳﺗﺧدم ّ‬‫ﻋﻠﯾك ﺣل اﻟﻣﺷﻛﻠﺔ ﺑﺎﺗﺑﺎع إرﺷﺎدات اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل اﻟﻣﻌﺗﻣد ﻓﻲ ﻣﻧطﻘﺗك أو‬‫ﺳﯾزود ﻗﺳم ﺧدﻣﺔ اﻟﻌﻣﻼء اﻟﺗﺎﺑﻊ ﻟﺷرﻛﺔ ‪ A&D‬اﻟﻣﻌﻠوﻣﺎت اﻟﻔﻧﯾﺔ وﻗطﻊ اﻟﻐﯾﺎر‬‫ﻗﺳم ﺧدﻣﺔ اﻟﻌﻣﻼء‪.‬‬‫ّ‬‫واﻟوﺣدات ﻟﻠوﻛﻼء اﻟﻣﻌﺗﻣدﯾن‪.‬‬‫ﺻﻧّﻊ ﻟﯾﺗﻣﺗﻊ ﺑﻔﺗرة ﺗﺷﻐﯾل طوﯾﻠﺔ‪ .‬وﻣﻊ ذﻟك‪ ،‬ﻓﺈﻧﮫ ﯾوﺻﻰ ﺑﺷﻛل ﻋﺎم ﺑﻔﺣص اﻟﺟﮭﺎز ﻛل‬‫ﺻﻣم اﻟﺟﮭﺎز و ُ‬‫ُ‬‫ﻋﺎﻣﯾن ﻟﺿﻣﺎن اﻟﺗﺷﻐﯾل اﻟﺻﺣﯾﺢ ﻟﮫ ودﻗﺔ أداﺋﮫ‪ .‬ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑﺎﻟوﻛﯾل اﻟﻣﻌﺗﻣد ﻓﻲ ﻣﻧطﻘﺗك أو ﺷرﻛﺔ‬‫‪ A&D‬ﻹﺟراء اﻟﺻﯾﺎﻧﺔ‪.‬‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻔﻧﯾﺔ‬‫اﻟﻧوع‬‫طرﯾﻘﺔ اﻟﻘﯾﺎس‬‫ﻧطﺎق اﻟﻘﯾﺎس‬‫دﻗﺔ اﻟﻘﯾﺎس‬‫ﻣﺻدر اﻹﻣداد ﺑﺎﻟطﺎﻗﺔ‬‫ﻋدد ﻣرات اﻟﻘﯾﺎس‬‫اﻟﺗﺻﻧﯾف‬‫‪UA-767S‬‬‫اﻟﻘﯾﺎس ﺑﺎﻟذﺑذﺑﺎت‬‫‪mmHg 299 - 0‬‬‫اﻟﺿﻐط‪:‬‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪mmHg 279 - 60 :‬‬‫اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪mmHg 200 - 40 :‬‬‫‪ 180 - 40‬ﻧﺑﺿﺔ‪ /‬اﻟدﻗﯾﻘﺔ‬‫اﻟﻧﺑض‪:‬‬‫‪mmHg 3±‬‬‫اﻟﺿﻐط‪:‬‬‫‪%5±‬‬‫اﻟﻧﺑض‪:‬‬‫‪ 4‬ﺑطﺎرﯾﺎت ﺑﺟﮭد ‪ 1.5‬ﻓوﻟت )ﻣﻘﺎس ‪ ،R6P‬أو ‪ ،LR6‬أو ‪(AA‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد )‪) (TB-233C‬ﻏﯾر ﻣرﻓﻖ(‬‫ّ‬‫اﻟﺑطﺎرﯾﺔ ﻣﻘﺎس ‪) LR6‬ﺑطﺎرﯾﺎت ﻗﻠوﯾﺔ(‬‫‪ 700‬ﻣرة ﻗﯾﺎس ﺗﻘرﯾﺑًﺎ‬‫اﻟﺑطﺎرﯾﺔ ﻣﻘﺎس ‪) R6P‬ﺑطﺎرﯾﺎت ﻣﻧﺟﻧﯾز(‬‫‪ 200‬ﻣرة ﻗﯾﺎس ﺗﻘرﯾﺑًﺎ‬‫ﻣﻊ ﻗﯾﻣﺔ ﺿﻐط ﺗﺳﺎوي ‪ mmHg 180‬ﻓﻲ ﻏرﻓﺔ درﺟﺔ ﺣرارﺗﮭﺎ‬‫‪ °23‬ﻣﺋوﯾﺔ‪.‬‬‫ﺟﮭﺎز طﺑﻲ ﻛﮭرﺑﺎﺋﻲ ﺑﻣﺻدر طﺎﻗﺔ داﺧﻠﻲ )ﯾﻌﻣل ﺑﺑطﺎرﯾﺎت( ‪ /‬اﻟﻔﺋﺔ ‪II‬‬‫ﺑﻣﺣول اﻟﺗﯾﺎر(‬‫)ﯾﻌﻣل‬‫ّ‬‫وﺿﻊ اﻟﺗﺷﻐﯾل اﻟﻣﺳﺗﻣر‬‫اﻻﺧﺗﺑﺎر اﻟﺳرﯾري‬‫وﻓﻘًﺎ ﻟﺷﮭﺎدة ‪ISO81060-2：2013‬‬‫‪EMD‬‬‫اﻟذاﻛرة‬‫ظروف اﻟﺗﺷﻐﯾل‬‫ظروف اﻟﻧﻘل ‪ /‬اﻟﺗﺧزﯾن‬‫ﻓﻲ دراﺳﺔ اﻟﺗﺣﻘﻖ ﻣن ﺻﺣﺔ اﻟﺗﺟﺎرب اﻟﺳرﯾرﯾﺔ‪ ،‬ﻛﺎن ‪ K5‬ﯾﺳﺗﺧدم‬‫ﻋﺎ ﻟﺗﺣدﯾد ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ‪.‬‬‫ﻓﻲ ‪ 85‬ﻣوﺿو ً‬‫‪IEC 60601-1-2: 2014‬‬‫آﺧر ‪ 60‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس‬‫‪ +10‬إﻟﻰ ‪ 15 / °C +40‬إﻟﻰ ‪ 800 / RH% 85‬إﻟﻰ ‪hPa 1,060‬‬‫‪ -20‬إﻟﻰ ‪ 10 / °C +60‬إﻟﻰ ‪ 700 / RH% 95‬إﻟﻰ ‪hPa 1,060‬‬‫اﻟﻌرﺑﯾﺔ ‪21‬‬‫اﻷﺑﻌﺎد‬‫اﻟوزن‬‫اﻟﺣﻣﺎﯾﺔ ﻣن اﻟﻣواد اﻟدﺧﯾﻠﺔ‬‫ﺗﻘرﯾﺑًﺎ ‪] 140‬ﻋرض[ × ‪] 60‬ارﺗﻔﺎع[ × ‪] 105‬ﻋﻣﻖ[ ﻣم‬‫ﺗﻘرﯾﺑًﺎ ‪ 245‬ﺟرا ًﻣﺎ دون اﻟﺑطﺎرﯾﺎت‬‫‪IP20‬‬‫اﻟﺟﮭﺎز‪:‬‬‫اﻟﻧوع ‪BF‬‬‫اﻟﺟزء اﻟﻣﻠﻔوف‬‫اﻟرﺑﺎط‬‫ﻣدة اﻟﺻﻼﺣﯾﺔ‬‫‪ 5‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫اﻟﺟﮭﺎز‪:‬‬‫ﺳﻧﺗﺎن )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫اﻟرﺑﺎط‪:‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪ 5 :‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻣﻠﺣﻖ ّ‬‫ﻣﺻدرا ﻟﻺﻣداد ﺑﺎﻟطﺎﻗﺔ ﻓﻲ اﻟﻣﻧزل‪ .‬ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑوﻛﯾل ‪ A&D‬اﻟﻣﺣﻠﻲ‬‫اﻟﻣﺣول ﺑﺎﻟﺟﮭﺎز ﻟﯾﻛون‬‫ﯾﺗﺻل‬‫ً‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد أو اﺳﺗﺑداﻟﮫ ﺑﺷﻛل دوري‪.‬‬‫ﻟﺷراﺋﮫ‪ .‬ﯾﻛون ﺿرورﯾًﺎ ﻓﺣص ّ‬‫ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑوﻛﯾل ‪ A&D‬اﻟﻣﺣﻠﻲ ﻟﺷراﺋﮫ‪.‬‬‫‪TB-233C‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد أو اﺳﺗﺑداﻟﮫ ﺑﺷﻛل دوري‪.‬‬‫ﯾﻛون ﺿرورﯾًﺎ ﻓﺣص ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ ّ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫اﻟرﻣوز‬‫ﻟﻼﺳﺗﺧدام اﻟﻣﻧزﻟﻲ ﻓﻘط‬‫اﻟﺟﮭﺎز ﻣن اﻟﻔﺋﺔ ‪II‬‬‫اﻟﻣﺻﮭر اﻟﺣراري‬‫اﻟﻣﺻﮭر‬‫ﻣﻠﺻﻖ اﻟﺗوﺟﯾﮫ ‪ EC‬ﻟﻸﺟﮭزة‬‫ﻣﻠﺻﻖ ﺷﮭﺎدة ‪ EAC‬ﻟﻸﺟﮭزة‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟﻘطﺑﯾﺔ ﻟﻘﺎﺑس ّ‬‫اﻟﻣﻠﺣﻘﺎت ﺗُﺑﺎع ﺑﺷﻛل ﻣﻧﻔﺻل‬‫ﺣﺟم اﻟرﺑﺎط‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫اﻟرﺑﺎط‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬‫‪CUF-F-LA‬‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬‫‪CUF-I‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪CUF-F-A‬‬‫ﺣﺟم اﻟذراع‪ :‬اﻟﻣﺣﯾط ﻋﻧد ﻋﺿﻠﺔ اﻟذراع‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ّ‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫‪TB-233C‬‬‫ﺣﺟم اﻟذراع‬‫‪ 31‬ﺳم إﻟﻰ ‪ 45‬ﺳم‬‫‪ 22‬ﺳم إﻟﻰ ‪ 42‬ﺳم‬‫‪ 22‬ﺳم إﻟﻰ ‪ 32‬ﺳم‬‫اﻟﻘﺎﺑس )ﻧوع اﻟﻣﺧرج(‬‫اﻟﻧوع ‪C‬‬

‫ﻣﻼﺣظﺔ‪ :‬ﺗﺧﺿﻊ اﻟﻣواﺻﻔﺎت ﻟﻠﺗﻐﯾﯾر دون إﺷﻌﺎر ﻣﺳﺑﻖ‪.‬‬‫ﺗﺻﻧﯾف ‪ IP‬ھو درﺟﺎت اﻟﺣﻣﺎﯾﺔ اﻟﺗﻲ ﯾﺗﻣﺗﻊ ﺑﮭﺎ اﻟﺟﮭﺎز وﻓﻘًﺎ ﻟﻠﺗوﺟﯾﮫ ‪ .IEC 60529‬ھذا اﻟﺟﮭﺎز‬‫ﻣﺣﻣﻲ ﺿد اﻟﻣواد اﻟﻐرﯾﺑﺔ اﻟﺻﻠﺑﺔ اﻟﺗﻲ ﯾﺑﻠﻎ ﻗطرھﺎ ‪ 12‬ﻣم أو أﻛﺑر‪ ،‬ﻣﺛل اﻷﺻﺎﺑﻊ‪ .‬ھذا اﻟﺟﮭﺎز ﻏﯾر‬‫ﻣﺣﻣﻲ ﺿد اﻟﻣﺎء‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪22‬‬‫ﻓﮭرﺳت ﻣﻧدرﺟﺎت‬‫ﻣﺷﺗرﯾﺎن ﮔراﻣﯽ ‪2 .......................................................................................‬‬‫ﺗذﮐرات اوﻟﯾﮫ‪2 ...........................................................................................‬‬‫اﻗداﻣﺎت اﺣﺗﯾﺎطﯽ ‪2 ......................................................................................‬‬‫ﻣﻌرﻓﯽ ﻗطﻌﺎت ‪5 .........................................................................................‬‬‫ﻧﻣﺎدھﺎ‪6 ... ................................................................................................‬‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت ‪8 ..........................................................................................‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش ‪9 ...............................................................................‬‬‫ﻧﺻب ‪ /‬ﺗﻌوﯾض ﺑﺎﺗریھﺎ‪9 ........................................................................‬‬‫ﻣﺗﺻل ﮐردن ﺷﯾﻠﻧﮓ ھوا ‪10 ......................................................................‬‬‫ﻣﺗﺻل ﮐردن آداﭘﺗور ‪10 ..... ................................................................AC‬‬‫اﻧﺗﺧﺎب ﺳﺎﯾز ﻣﻧﺎﺳب ﺑﺎزوﺑﻧد ‪11 .. ................................................................‬‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد ‪11 ..............................................................................‬‬‫ﻧﺣوه اﻧﺟﺎم اﻧدازهﮔﯾری دﻗﯾﻖ ‪13 .. ................................................................‬‬‫اﻧدازهﮔﯾری ‪13 .....................................................................................‬‬‫ﭘس از اﻧدازهﮔﯾری ‪13 .............................................................................‬‬‫اﻧدازهﮔﯾریھﺎ ‪14 ........................................................................................‬‬‫اﻧدازهﮔﯾری ﻋﺎدی ‪14 ..............................................................................‬‬‫اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‪15 ........................................................‬‬‫ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ ‪15 . ................................................................‬‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ ‪16 .........................................................................‬‬‫ﺷﺎﺧص ‪ IHB/AFib‬ﭼﯾﺳت؟ ‪17 ......................................................................‬‬‫‪ AFib‬ﭼﯾﺳت؟ ‪17 ......................................................................................‬‬‫‪17 ..................................................................................... ％IHB/AFib‬‬‫ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر ‪18 ...................................................................................‬‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪18 ............................................(WHO‬‬‫در ﻣورد ﻓﺷﺎر ﺧون‪18 .................................................................................‬‬‫ﻓﺷﺎر ﺧون ﭼﯾﺳت؟‪18 .............................................................................‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﭼﯾﺳت و ﭼﮕوﻧﮫ ﮐﻧﺗرل ﻣﯽﺷود؟ ‪19 ............................................‬‬‫ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ اﻧدازهﮔﯾری ﮐﻧم؟ ‪19 ................................................‬‬‫طﺑﻘﮫﺑﻧدی ﻓﺷﺎر ﺧون ‪19 . ................................................................ WHO‬‬‫ﻧوﺳﺎﻧﺎت ﻓﺷﺎر ﺧون‪19 ............................................................................‬‬‫ﻋﯾبﯾﺎﺑﯽ ‪20 .............................................................................................‬‬‫ﻧﮕﮭداری ‪21 .............................................................................................‬‬‫دادهھﺎی ﻓﻧﯽ ‪21 .........................................................................................‬‬‫ﻓﺎرﺳﯽ ‪1‬‬‫ﻣﺷﺗرﯾﺎن ﮔراﻣﯽ‬‫ﺑﺎﺑت ﺧرﯾد ﺟدﯾدﺗرﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ‪ A&D‬ﮐﮫ ﯾﮑﯽ از ﭘﯾﺷرﻓﺗﮫﺗرﯾن دﺳﺗﮕﺎهھﺎی ﭘﺎﯾش ﻣوﺟود‬‫در ﺑﺎزار اﺳت‪ ،‬ﺑﮫ ﺷﻣﺎ ﺗﺑرﯾﮏ ﻣﯽﮔوﯾﯾم‪ .‬اﯾن دﺳﺗﮕﺎه ﮐﮫ ﺑرای اﺳﺗﻔﺎده آﺳﺎن و در ﻋﯾن ﺣﺎل دﻗﯾﻖ طراﺣﯽ‬‫ﺷده اﺳت‪ ،‬ﺑررﺳﯽ روزاﻧﮫ ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﺳﮭﯾل ﻣﯽﮐﻧد‪.‬‬‫ﺗوﺻﯾﮫ ﻣﯽﺷود ﭘﯾش از آﻧﮑﮫ ﺑرای اوﻟﯾن ﺑﺎر از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬اﯾن دﻓﺗرﭼﮫ راھﻧﻣﺎ را‬‫ﺑﮫدﻗت ﻣطﺎﻟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﺗذﮐرات اوﻟﯾﮫ‬‫اﯾن دﺳﺗﮕﺎه ﺑﺎ دﺳﺗوراﻟﻌﻣل ‪ 93/42 EEC‬ﺑرای ﻣﺣﺻوﻻت ﭘزﺷﮑﯽ در اروﭘﺎ ﻣطﺎﺑﻘت دارد‪ .‬ﻧﺷﺎن‬‫اﻧطﺑﺎق‬‫ﺷﺎھدی ﺑر اﯾن ادﻋﺎ اﺳت‪ :0123) .‬ﺷﻣﺎره ارﺟﺎع ﺑﮫ ﺑﺧش اﻋﻼم ﺷده ﻣرﺑوطﮫ(‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﺳﺗﻔﺎده ﺑزرﮔﺳﺎﻻن طراﺣﯽ ﺷده اﺳت‪ ،‬ﻧﮫ ﻧوزادان ﯾﺎ اطﻔﺎل‪.‬‬‫ﻣﺣﯾط ﮐﺎرﺑرد‪ .‬اﯾن دﺳﺗﮕﺎه ﺑﮫﮔوﻧﮫای طراﺣﯽ ﺷده اﺳت ﮐﮫ ﺷﻣﺎ ﺑﺗواﻧﯾد در ﻣﺣﯾط ﻣﻧزﻟﺗﺎن ﺷﺧﺻﺎ ً از آن‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون و ﺿرﺑﺎن ﻗﻠب اﻓراد ﺑرای ﺗﺷﺧﯾص ﺑﯾﻣﺎری طراﺣﯽ ﺷده اﺳت‪.‬‬‫اﻗداﻣﺎت اﺣﺗﯾﺎطﯽ‬‫در ﺳﺎﺧت اﯾن دﺳﺗﮕﺎه از اﺟزای دﻗﯾﻖ اﺳﺗﻔﺎده ﺷده اﺳت‪ .‬دﻣﺎ و رطوﺑت ﺷدﯾد‪ ،‬ﻧور ﻣﺳﺗﻘﯾم ﺧورﺷﯾد‪،‬‬‫ﺿرﺑﮫ ﯾﺎ ﮔرد و ﺧﺎک ﻣﯽﺗواﻧد ﺑﮫ آن آﺳﯾب ﺑرﺳﺎﻧد‪.‬‬‫ﺑﺎزوﺑﻧد و دﺳﺗﮕﺎه را ﺑﺎ ﯾﮏ ﭘﺎرﭼﮫ ﻧرم و ﺧﺷﮏ ﯾﺎ ﭘﺎرﭼﮫ ﻧﻣدار ﺷده ﺑﺎ آب ﯾﺎ ﺷوﯾﻧده ﺧﻧﺛﯽ ﺗﻣﯾز ﮐﻧﯾد‪.‬‬‫ھرﮔز از اﻟﮑل‪ ،‬ﺑﻧزن‪ ،‬ﺗﯾﻧر ﯾﺎ دﯾﮕر ﻣواد ﺷﯾﻣﯾﺎﯾﯽ ﺧورﻧده ﺑرای ﺗﻣﯾز ﮐردن ﺑﺎزوﺑﻧد ﯾﺎ دﺳﺗﮕﺎه اﺳﺗﻔﺎده‬‫ﻧﮑﻧﯾد‪.‬‬‫از ﺧم ﮐردن زﯾﺎد ﺑﺎزوﺑﻧد ﯾﺎ ﭘﯾﭼﺎﻧدن ﺷﯾﻠﻧﮓ ﺑرای ﻣدت طوﻻﻧﯽ اﺟﺗﻧﺎب ﮐﻧﯾد‪ ،‬زﯾرا اﯾن ﮐﺎر ﻣوﺟب ﮐوﺗﺎه‬‫ﺷدن ﻋﻣر اﺟزای دﺳﺗﮕﺎه ﻣﯽﺷود‪.‬‬‫ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ ﺷﯾﻠﻧﮓ و ﮐﺎﺑل دﺳﺗﮕﺎه ﺑﮫ دور ﮔردن ﮐودﮐﺎن ﯾﺎ ﻧوزادان ﻧﭘﯾﭼد‪ ،‬زﯾرا ﻣﯽﺗواﻧد ﻣوﺟب‬‫ﺧﻔﮕﯽ آﻧﮭﺎ ﺷود‪.‬‬‫ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ در ﺣﯾن اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺷﯾﻠﻧﮓ ھوا دﭼﺎر ﭘﯾﭻﺧوردﮔﯽ ﻧﺷود‪ .‬اﯾن اﻣر ﻣﯽﺗواﻧد ﺑر‬‫اﺛر ﻓﺷﺎر ﭘﯾوﺳﺗﮫ ﺑﺎزوﺑﻧد ﺑﺎﻋث اﯾﺟﺎد آﺳﯾب ﺷود‪.‬‬‫دﺳﺗﮕﺎه و ﺑﺎزوﺑﻧد در ﺑراﺑر آب ﻣﻘﺎوم ﻧﯾﺳﺗﻧد‪ .‬ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ آب‪ ،‬ﺑﺎران و ﻋرق روی دﺳﺗﮕﺎه و ﺑﺎزوﺑﻧد‬‫ﻟﮑﮫ اﯾﺟﺎد ﻧﮑﻧﻧد‪.‬‬‫اﮔر از دﺳﺗﮕﺎه در ﻧزدﯾﮑﯽ ﺗﻠوﯾزﯾون‪ ،‬ﻓر ﻣﺎﯾﮑرووﯾو‪ ،‬ﺗﻠﻔن ھﻣراه‪ ،‬اﺷﻌﮫ اﯾﮑس ﯾﺎ در ﻧزدﯾﮑﯽ دﯾﮕر‬‫اﺑزارھﺎﯾﯽ اﺳﺗﻔﺎده ﮐﻧﯾد ﮐﮫ ﻣﯾدان اﻟﮑﺗرﯾﮑﯽ ﻗوی دارﻧد‪ ،‬ﻣﻣﮑن اﺳت در اﻧدازهﮔﯾری اﺧﺗﻼل اﯾﺟﺎد ﺷود‪.‬‬‫دﺳﺗﮕﺎهھﺎی ارﺗﺑﺎطﯽ ﺑﯽﺳﯾم ﻣﺎﻧﻧد دﺳﺗﮕﺎهھﺎی ﺷﺑﮑﮫ ﺧﺎﻧﮕﯽ و ﺗﻠﻔنھﺎی ھﻣراه‪ ،‬ﺗﻠﻔنھﺎی ﺑﯽﺳﯾم و ﭘﺎﯾﮫ‬‫آﻧﮭﺎ‪ ،‬و ﺑﯾﺳﯾمھﺎ ﻣﯽﺗواﻧﻧد ﺑر اﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺗﺄﺛﯾر ﺑﮕذارﻧد‪ .‬ﻟذا ﻻزم اﺳت ﮐﮫ ﺣداﻗل‬‫‪ 30‬ﺳﺎﻧﺗﯽ ﻣﺗر از اﯾن دﺳﺗﮕﺎهھﺎ ﻓﺎﺻﻠﮫ داﺷﺗﮫ ﺑﺎﺷﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪2‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد ﻣﺟددا ً از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬از ﺗﻣﯾز ﺑودن دﺳﺗﮕﺎه ﻣطﻣﺋن ﺷوﯾد‪.‬‬‫ﺗﺟﮭﯾزات‪ ،‬ﻗطﻌﺎت و ﺑﺎﺗریھﺎﯾﯽ ﮐﮫ اﺳﺗﻔﺎده ﺷدهاﻧد‪ ،‬ﻧﺑﺎﯾد ﺑﮫﻋﻧوان ﺿﺎﯾﻌﺎت ﻋﺎدی ﺧﺎﻧﮕﯽ ﻗﻠﻣداد ﺷوﻧد‪ ،‬و‬‫ﺑﺎﯾد ﺑر اﺳﺎس ﻣﻘررات ﻣﺣﻠﯽ دور اﻧداﺧﺗﮫ ﺷوﻧد‪.‬‬‫وﻗﺗﯽ از آداﭘﺗور ‪ AC‬اﺳﺗﻔﺎده ﻣﯽﮐﻧﯾد‪ ،‬ﻣطﻣﺋن ﺷوﯾد ﮐﮫ آداﭘﺗور ‪ AC‬ﻣﯽﺗواﻧد در ﺻورت ﻟزوم ﺑﮫ‬‫ﺳﮭوﻟت از ﭘرﯾز ﺑرق ﺧﺎرج ﺷود‪.‬‬‫ﺗﻐﯾﯾری در دﺳﺗﮕﺎه اﯾﺟﺎد ﻧﮑﻧﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺑﮫ دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﺑﮫﻣﻧظور اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪ ،‬ﺑﺎزو ﺑﺎﯾد ﺗوﺳط ﺑﺎزوﺑﻧد ﺑﮫﻗدری ﻓﺷرده ﺷود ﮐﮫ ﻣوﻗﺗﺎ ً ﺟرﯾﺎن ﺧون از‬‫ﻣﯾﺎن ﺳرﺧرگ را ﻣﺗوﻗف ﮐﻧد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد درد‪ ،‬ﮐرﺧﺗﯽ ﯾﺎ ﻗرﻣزی ﻣوﻗت ﺑﺎزو ﺷود‪.‬‬‫ﭼﻧﯾن ﺷراﯾطﯽ ﺑﮫوﯾژه زﻣﺎﻧﯽ اﯾﺟﺎد ﻣﯽﺷود ﮐﮫ اﻧدازهﮔﯾری را ﭼﻧد ﻣرﺗﺑﮫ ﭘﺷت ﺳرھم اﻧﺟﺎم ﻣﯽدھﯾد‪.‬‬‫ھرﮔوﻧﮫ درد‪ ،‬ﮐرﺧﺗﯽ ﯾﺎ ﻗرﻣزی ﺑﺎ ﮔذﺷت زﻣﺎن از ﺑﯾن ﻣﯽرود‪.‬‬‫اﮔر اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﭼﻧدﯾن ﻣرﺗﺑﮫ ﺗﮑرار ﺷود‪ ،‬ﻣﯽﺗواﻧد ﺑﮫ دﻟﯾل ﺗداﺧﻠﯽ ﮐﮫ در ﺟرﯾﺎن ﺧون اﯾﺟﺎد‬‫ﻣﯽﮐﻧد‪ ،‬ﺑﮫ ﻓرد آﺳﯾب ﺑزﻧد‪ .‬اﯾن ﻣوﺿوع را ﺑررﺳﯽ ﮐﻧﯾد ﮐﮫ در ﺻورت ﺗﮑرار اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪،‬‬‫ﻋﻣﻠﯾﺎت دﺳﺗﮕﺎه ﻣوﺟب اﺧﺗﻼل طوﻻﻧﯽ ﻣدت در ﺟرﯾﺎن ﺧون ﻧﺷود‪.‬‬‫اﮔر ﻋﻣل ﻣﺎﺳﺗﮑﺗوﻣﯽ داﺷﺗﮫاﯾد و ﺳﯾﻧﮫﺗﺎن ﺗﺧﻠﯾﮫ ﺷده اﺳت‪ ،‬ﭘﯾش از اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺑﺎ ﭘزﺷﮑﺗﺎن‬‫ﻣﺷورت ﮐﻧﯾد‪.‬‬‫اﺟﺎزه ﻧدھﯾد ﮐﮫ ﮐودﮐﺎن از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﻧد و در ﺟﺎﯾﯽ ﮐﮫ در دﺳﺗرس اطﻔﺎل اﺳت از اﯾن دﺳﺗﮕﺎه‬‫اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪ .‬ﻋدم ﺗوﺟﮫ ﺑﮫ اﯾن ﻣوﺿوع ﻣﯽﺗواﻧد ﺑﺎﻋث ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺷود‪.‬‬‫اﺟزای ﮐوﭼﮑﯽ در اﯾن دﺳﺗﮕﺎه وﺟود دارد ﮐﮫ در ﺻورت ﺑﻠﻌﯾده ﺷدن ﺗوﺳط اطﻔﺎل ﻣﯽﺗواﻧد ﺑﺎﻋث ﺧﻔﮕﯽ‬‫آﻧﮭﺎ ﺷود‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ دﯾﮕر ﻧﻣﯽﺧواھﯾد از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬آداﭘﺗور ‪ AC‬را ﺧﺎرج ﮐﻧﯾد‪.‬‬‫اﺳﺗﻔﺎده از ﻟوازم ﺟﺎﻧﺑﯽ ﮐﮫ در اﯾن راھﻧﻣﺎ ﺑﮫ آﻧﮭﺎ اﺷﺎره ﻧﺷده اﺳت ﻣﯽﺗواﻧد ﺑﺎﻋث آﺳﯾب ﺑﮫ ﻓرد و دﺳﺗﮕﺎه‬‫ﺷود‪.‬‬‫در ﺻورﺗﯽ ﮐﮫ ﺑﺎﺗری اﺗﺻﺎل ﮐوﺗﺎه ﺷود‪ ،‬ﻣﻣﮑن اﺳت داغ ﺷده و ﺑﺎﻋث اﯾﺟﺎد ﺳوﺧﺗﮕﯽ ﺷود‪.‬‬‫ﭘﯾش از اﺳﺗﻔﺎده )ﺣدود ﯾﮏ ﺳﺎﻋت(‪ ،‬اﺟﺎزه دھﯾد ﮐﮫ دﺳﺗﮕﺎه ﺧودش را ﺑﺎ ﻣﺣﯾط اطراف ﺳﺎزﮔﺎر ﮐﻧد‪.‬‬‫آزﻣﺎﯾش ﺑﺎﻟﯾﻧﯽ روی ﻧوزادان و زﻧﺎن ﺑﺎردار اﻧﺟﺎم ﻧﺷده اﺳت‪ .‬ﻟذا از اﯾن دﺳﺗﮕﺎه ﺑرای ﻧوزادان و زﻧﺎن‬‫ﺑﺎردار اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ‪ ،‬ﻓﯾش ‪ DC‬و ﺧود ﺑﯾﻣﺎر را ھﻣزﻣﺎن ﻟﻣس ﻧﮑﻧﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺑرقﮔرﻓﺗﮕﯽ‬‫ﺷود‪.‬‬‫از ﺑﺎد ﮐردن دﺳﺗﮕﺎه ﺑدون ﭘﯾﭼﺎﻧدن آن ﺑﮫ دور ﺑﺎزو اﺟﺗﻧﺎب ﮐﻧﯾد‪.‬‬‫ﻣوارد ﻣﻧﻊ اﺳﺗﻔﺎده‬‫در اﯾن ﺑﺧش ﻣوارد اﺣﺗﯾﺎطﯽ ﺑرای اﺳﺗﻔﺎده ﺻﺣﯾﺢ از دﺳﺗﮕﺎه ذﮐر ﺷده اﺳت‪.‬‬‫ﺑﺎزوﺑﻧد را روی ﺑﺎزوﯾﯽ ﮐﮫ ﺗﺟﮭﯾزات اﻟﮑﺗرﯾﮑﯽ ﭘزﺷﮑﯽ دﯾﮕری روی آن ﻣﺗﺻل اﺳت‪ ،‬ﺑﮫ ﮐﺎر ﻧﺑرﯾد‪ .‬در‬‫ﭼﻧﯾن ﺷراﯾطﯽ ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫﺧوﺑﯽ ﮐﺎر ﻧﮑﻧد‪.‬‬‫اﻓرادی ﮐﮫ دﭼﺎر ﻧﻘص ﺷدﯾد ﮔردش ﺧون در ﺑﺎزوھﺎﯾﺷﺎن ھﺳﺗﻧد ﺑﺎﯾد ﭘﯾش از اﺳﺗﻔﺎده از اﯾن دﺳﺗﮕﺎه ﺑﺎ‬‫ﭘزﺷﮑﺷﺎن ﻣﺷورت ﮐﻧﻧد ﺗﺎ از اﯾﺟﺎد ﻣﺷﮑﻼت ﭘزﺷﮑﯽ اﺟﺗﻧﺎب ﺷود‪.‬‬‫از ﺗﺷﺧﯾص ﺧودﺳراﻧﮫ ﺑر اﺳﺎس ﻧﺗﺎﯾﺞ اﻧدازهﮔﯾری ﺑﭘرھﯾزﯾد و از ھرﮔوﻧﮫ ﺧوددرﻣﺎﻧﯽ اﺟﺗﻧﺎب ﮐﻧﯾد‪.‬‬‫ھﻣﯾﺷﮫ ﺑرای ارزﯾﺎﺑﯽ ﻧﺗﯾﺟﮫ و ﺗﺟوﯾز درﻣﺎن ﻣﻧﺎﺳب ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪3‬‬‫ﺑﺎزوﺑﻧد دﺳﺗﮕﺎه ﻓﺷﺎر ﺧون را روی ﺑﺎزوﯾﯽ ﮐﮫ زﺧم درﻣﺎن ﻧﺷده دارد ﻗرار ﻧدھﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را روی ﺑﺎزوﯾﯽ ﮐﮫ ﺗﺣت ﺗزرﯾﻖ ورﯾدی ﯾﺎ اﻧﺗﻘﺎل ﺧون اﺳت‪ ،‬ﻗرار ﻧدھﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد‬‫ﺑﺎﻋث ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺷود‪.‬‬‫در ﺟﺎﯾﯽ ﮐﮫ ﮔﺎزھﺎی اﺷﺗﻌﺎلﭘذﯾر ﻣﺎﻧﻧد ﮔﺎزھﺎی ﺑﯾﮭوﺷﯽ وﺟود دارد ﻧﺑﺎﯾد از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﺷود‪ .‬اﯾن‬‫ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﻧﻔﺟﺎر ﺷود‪.‬‬‫در ﻣﺣﯾطھﺎﯾﯽ ﻣﺎﻧﻧد اﺗﺎﻗﮏ اﮐﺳﯾژن ﻓﺷﺎر ﺑﺎﻻ ﯾﺎ ﭼﺎدر اﮐﺳﯾژن ﮐﮫ اﮐﺳﯾژن ﺑﺳﯾﺎر ﻣﺗراﮐم دارﻧد‪ ،‬ﻧﺑﺎﯾد از‬‫اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﺷود‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث آﺗشﺳوزی ﯾﺎ اﻧﻔﺟﺎر ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪4‬‬‫ﻣﻌرﻓﯽ ﻗطﻌﺎت‬‫ﻧﻣﺎﯾﺷﮕر‬‫ﻧری راﺑط ھوا‬‫ﻧری آداﭘﺗور ‪AC‬‬‫ﻓﯾش ‪DC‬‬‫دﮐﻣﮫ ‪START‬‬‫ﻣﺎدﮔﯽ ھوا‬‫ﺷﯾﻠﻧﮓ ھوا‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﻧﺷﺎﻧﮫ ﺷﺎﺧص‬‫ﺑﺎزوﺑﻧد‬‫ﺣﺎﻓظﮫ‬‫درﭘوش ﺑﺎﺗری‬‫ﺑﺎﺗریھﺎی ‪ 1.5‬وﻟت‬‫)‪ LR6 ،R6P‬ﯾﺎ ‪(AA‬‬‫ﻧﻣﺎﯾﺷﮕر‬‫ﻣﯾﺎﻧﮕﯾن‬‫ﺣﺎﻓظﮫ‬‫ﻧﻣﺎد ‪IHB/AFib‬‬‫ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ‬‫‪%IHB/AFib‬‬‫ﻧﻣﺎد ﺧطﺎی اﻧدازه ﺑﺎزوﺑﻧد‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫ﻧﻣﺎد ﺧطﺎی ﻧﺎﺷﯽ از ﺣرﮐت‬‫ﻋﻼﻣت ﻗﻠب‬‫ﺿرﺑﺎن ﻧﺑض‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ‬‫)‪ (WHO‬و ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر‬‫ﺷﺎﺧص ﺑﺎﺗری‬‫ﻓﺎرﺳﯽ ‪5‬‬‫ﻧﻣﺎدھﺎ‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی ﺟﻌﺑﮫ دﺳﺗﮕﺎه ﭼﺎپ ﺷدهاﻧد‬‫ﻧﻣﺎدھﺎ‬‫‪SYS‬‬‫‪DIA‬‬‫‪PUL‬‬‫‪IP‬‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﺣﺎﻟت آﻣﺎدهﺑﮫﮐﺎر و روﺷن ﮐردن دﺳﺗﮕﺎه‪.‬‬‫ﻓﺷﺎر ﺧون ﺳﯾﺳﺗوﻟﯽ ﺑر ﺣﺳب ‪mmHg‬‬‫ﻓﺷﺎر ﺧون دﯾﺎﺳﺗوﻟﯽ ﺑر ﺣﺳب ‪mmHg‬‬‫ﻧﺑض در دﻗﯾﻘﮫ‬‫راھﻧﻣﺎی ﻧﺻب ﺑﺎﺗری‬‫ﺟرﯾﺎن ﻣﺳﺗﻘﯾم‬‫ﻧوع ‪ :BF‬دﺳﺗﮕﺎه‪ ،‬ﺑﺎزوﺑﻧد و ﻟوﻟﮫھﺎ ﺑﮫﮔوﻧﮫای طراﺣﯽ ﺷدهاﻧد ﮐﮫ از ﺑروز ﺑرقﮔرﻓﺗﮕﯽ‬‫ﺟﻠوﮔﯾری ﺷود‪.‬‬‫ﺑرﭼﺳب دﺳﺗوراﻟﻌﻣل ‪ EC‬دﺳﺗﮕﺎه ﭘزﺷﮑﯽ‬‫ﻧﻣﺎﯾﻧده ‪EU‬‬‫ﺗوﻟﯾدﮐﻧﻧده‬‫ﺗﺎرﯾﺦ ﺗوﻟﯾد‬‫ﻧﻣﺎد ﺣﻔﺎظت ﺑﯾناﻟﻣﻠﻠﯽ‬‫ﺑرﭼﺳب ‪WEEE‬‬‫‪SN‬‬‫ﺷﻣﺎره ﺳ﷼‬‫ﺑﮫ ﮐﺗﺎﺑﭼﮫ‪/‬دﻓﺗرﭼﮫ راھﻧﻣﺎ ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫ﻗطﺑﯾت ﻓﯾش ‪DC‬‬‫در ﺟﺎی ﺧﺷﮏ ﻧﮕﮫداری ﮐﻧﯾد‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی ﻧﻣﺎﯾﺷﮕر ظﺎھر ﻣﯽﺷوﻧد‬‫ﻧﻣﺎدھﺎ‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم‬‫اﺳت‪ .‬ھﻧﮕﺎﻣﯽ ﮐﮫ ﻧﺑض ﺷﻧﺎﺳﺎﯾﯽ ﺷد‪ ،‬ﭼﺷﻣﮏ‬‫ﻣﯽزﻧد‪.‬‬‫ﻧﻣﺎد ‪IHB/AFib‬‬‫زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ ﺿرﺑﺎن ﻗﻠب ﻧﺎﻣﻧظم‬‫ﺷﻧﺎﺳﺎﯾﯽ ﺷود‪.‬‬‫اﯾن ﻧﻣﺎد زﻣﺎﻧﯽ ﮐﮫ ﺗﮑﺎنھﺎی ﺑﺳﯾﺎر آراﻣﯽ ﻣﺎﻧﻧد‬‫ﻟرزش ﺷﻧﺎﺳﺎﯾﯽ ﺷود ﻧﯾز ﻣﻣﮑن اﺳت روﺷن ﺷود‪.‬‬‫زﻣﺎﻧﯽ ظﺎھرا ً ﻣﯽﺷود ﮐﮫ ﺣرﮐﺗﯽ در ﺑدن ﯾﺎ ﺑﺎزو‬‫ﺷﻧﺎﺳﺎﯾﯽ ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪6‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‪ .‬ﺗﺎ‬‫ﺟﺎﯾﯽ ﮐﮫ ﻣﯽﺗواﻧﯾد ﺑﯽﺣرﮐت ﺑﻣﺎﻧﯾد‪.‬‬‫ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﻋدد ﻧﺎدرﺳﺗﯽ را‬‫ﻧﺷﺎن دھد‪.‬‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪ .‬در‬‫طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت ﺑﻣﺎﻧﯾد‪.‬‬‫ﻧﻣﺎدھﺎ‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫زﻣﺎﻧﯽ آﺷﮑﺎر ﻣﯽﺷود ﮐﮫ در ﺣﯾن اﻧدازهﮔﯾری‬‫ﺑﺎزوﺑﻧد ﺷل ﺑﺎﺷد‬‫ﻧرخ ﺷﻧﺎﺳﺎﯾﯽﺷده ‪ IHB/AFib‬در ﺣﺎﻓظﮫ‬‫‪= %IHB/AFib‬‬‫ﺗﻌداد ‪IHB/AFibs‬‬‫ﺷﻧﺎﺳﺎﯾﯽﺷده در ﺣﺎﻓظﮫ‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﻋدد ﻧﺎدرﺳﺗﯽ را‬‫ﻧﺷﺎن دھد‪ .‬ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ ﺑﺑﻧدﯾد‬‫و اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫‪[%] 100x‬‬‫ﺗﻌداد ﮐل‬‫اﻧدازهﮔﯾریھﺎی ﻗﺑﻠﯽ ﮐﮫ در ﺣﺎﻓظﮫ ذﺧﯾره ﺷده اﺳت‪.‬‬‫ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت‬‫‪) FULL BATTERY‬ﺑﺎﺗری ﭘر اﺳت(‬‫ﻧﺷﺎﻧﮕر ﺗوان ﺑﺎﺗری در طول اﻧدازهﮔﯾری‪.‬‬‫‪) LOW BATTERY‬ﺑﺎﺗری ﮐم(‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﺎﺗری ﺿﻌﯾف ﺷود‪ ،‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﻓﺷﺎر ﺧون ﺑﮫ دﻟﯾل ﺗﮑﺎن ﺧوردن در طول‬‫اﻧدازهﮔﯾری ﻧﺎﭘﺎﯾدار اﺳت‪.‬‬‫ﻣﻘﺎدﯾر ﺳﯾﺳﺗوﻟﯽ و دﯾﺎﺳﺗوﻟﯽ ﮐﻣﺗر از ‪mmHg 10‬‬‫ﺟﯾوه ﺑﺎ ﯾﮑدﯾﮕر اﺧﺗﻼف دارﻧد‪.‬‬‫ﻣﻘدار ﻓﺷﺎر در زﻣﺎن ﺑﺎد ﺷدن اﻓزاﯾش ﻧﯾﺎﻓت‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫدرﺳﺗﯽ ﻗرار ﻧﮕرﻓﺗﮫ اﺳت‪.‬‬‫‪PUL DISPLAY ERROR‬‬‫ﻧﺑض ﺑﮫدرﺳﺗﯽ ﺷﻧﺎﺳﺎﯾﯽ ﻧﺷده اﺳت‪.‬‬‫ﺧطﺎی داﺧﻠﯽ دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون‬‫ﻓﺎرﺳﯽ ‪7‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ اﯾن ﻧﺷﺎﻧﮫ ﭼﺷﻣﮏ ﻣﯽزﻧد‪،‬‬‫ھﻣﮫ ﺑﺎﺗریھﺎ را ﺗﻌوﯾض ﮐﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫در طول اﻧدازهﮔﯾری ﮐﺎﻣﻼً ﺑﯽﺣرﮐت‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ ﺑﺑﻧدﯾد‪ ،‬و‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ را ﺧﺎرج ﮐرده و دﮐﻣﮫ‬‫‪ START‬را ﺑزﻧﯾد‪ ،‬و ﺑﺎر دﯾﮕر‬‫ﺑﺎﺗریھﺎ را ﻧﺻب ﮐﻧﯾد‪ .‬اﮔر ﺧطﺎ‬‫ھﻣﭼﻧﺎن ظﺎھر ﺷد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﺗﻣﺎس‬‫ﺑﮕﯾرﯾد‪.‬‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت‬‫‪ .1‬اﻧدازهﮔﯾری ﻋﺎدی‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪ .‬ﻓﺷﺎر ﺧون اﻧدازهﮔﯾری ﺷده و اطﻼﻋﺎت در ﺣﺎﻓظﮫ ذﺧﯾره ﻣﯽﺷود‪ .‬اﯾن‬‫دﺳﺗﮕﺎه ﻣﯽﺗواﻧد ‪ 60‬اﻧدازهﮔﯾری آﺧر را در ﺣﺎﻓظﮫاش ذﺧﯾره ﮐﻧد‪.‬‬‫‪ .2‬ﻓراﺧواﻧﯽ اطﻼﻋﺎت‬‫در ﺣﺎﻟت آﻣﺎدهﺑﮫﮐﺎر‪،‬‬‫دﮐﻣﮫ را ﻓﺷﺎر دھﯾد و‬‫ﻧﮕﮫدارﯾد‬‫‪START‬‬‫اﮔر ھﯾﭻ ﭼﯾز ﻧﻣﺎﯾش داده ﻧﺷد‪ START ،‬را ﻓﺷﺎر داده و ﻧﮕﺎه‬‫دارﯾد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت ﻧﺷﺎن داده ﺷد‪ ،‬دﮐﻣﮫ را رھﺎ ﮐﻧﯾد‪.‬‬‫ﺷﻣﺎره اطﻼﻋﺎت و اطﻼﻋﺎت ذﺧﯾره ﺷده ﺑﮫﺻورت ﺧودﮐﺎر و ﺑﮫ‬‫ﺗرﺗﯾب از آﺧرﯾن اﻧدازهﮔﯾری ﻧﺷﺎن داده ﻣﯽﺷوﻧد‪.‬‬‫ﺑرای ﻣﺷﺎھده ﺟزﺋﯾﺎت ﺑﯾﺷﺗر در ﻣورد ﻓراﺧواﻧﯽ اطﻼﻋﺎت‪ ،‬ﺑﮫ‬‫ﺑﺧش "ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ" ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫‪ .3‬ﺣذف ﮐﻠﯾﮫ اطﻼﻋﺎت ذﺧﯾرهﺷده در ﺣﺎﻓظﮫ‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ دﺳﺗﮕﺎه را ﺧﺎﻣوش ﮐردﯾد‪ ،‬دﮐﻣﮫ‬‫‪ START‬را ﻓﺷﺎر داده و ﻧﮕﺎه دارﯾد ﺗﺎ زﻣﺎﻧﯽ ﮐﮫ‬‫" ظﺎھر ﺷود‪.‬‬‫"‬‫" را اﻧﺗﺧﺎب ﮐﻧﯾد ﺗﺎ اطﻼﻋﺎت ﭘﺎک ﺷود‪.‬‬‫"‬‫‪START‬‬‫ﭼﺷﻣﮏ ﺑزﻧد اطﻼﻋﺎت ﭘﺎک ﻣﯽﺷود‪.‬‬‫زﻣﺎﻧﯽ ﮐﮫ ﻋﻼﻣت‬‫دﺳﺗﮕﺎه ﺑﮫﺻورت ﺧودﮐﺎر ﺧﺎﻣوش ﻣﯽﺷود‪.‬‬‫‪ .4‬اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫ﺑﮫ ﺻﻔﺣﮫ ‪ 15‬ﻣراﺟﻌﮫ ﮐﻧﯾد ﺗﺎ اطﻼﻋﺎت ﺑﯾﺷﺗری در ﻣورد اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب ﺑﯾﺎﺑﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪8‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﺻب ‪ /‬ﺗﻌوﯾض ﺑﺎﺗریھﺎ‬‫‪ .1‬درﭘوش ﺑﺎطری را ﺑردارﯾد‪.‬‬‫‪ .2‬ﺑﺎﺗریھﺎی ﮐﮭﻧﮫ را ﺑردارﯾد و ھﻣﺎنطور ﮐﮫ ﻧﺷﺎن‬‫داده ﺷده اﺳت‪ ،‬ﺑﺎﺗریھﺎی ﻧو را در ﻣﺣﻔظﮫ‬‫ﺑﺎطری ﻗرار دھﯾد‪ .‬ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ ﻗطبھﺎی‬‫ﺑﺎﺗری )‪ +‬و ‪ (-‬در ﺟﺎی ﺻﺣﯾﺢ ﻗرار ﮔﯾرﻧد‪.‬‬‫ﺻرﻓﺎ ً از ﺑﺎﺗریھﺎی ‪ LR6 ،R6P‬ﯾﺎ ‪ AA‬اﺳﺗﻔﺎده‬‫ﮐﻧﯾد‪.‬‬‫‪ .3‬درﭘوش ﺑﺎﺗری را ﺑﮕذارﯾد‪.‬‬‫ﻣرﺣﻠﮫ ‪1‬‬‫ﻣرﺣﻠﮫ ‪3‬‬‫ﻣرﺣﻠﮫ ‪1‬‬‫ﻣرﺣﻠﮫ ‪2‬‬‫اﺣﺗﯾﺎط‬‫ﺑﺎﺗریھﺎ را طﺑﻖ ﺗﺻوﯾر در ﻣﺣﻔظﮫ ﺑﺎطری ﻗرار دھﯾد‪ .‬اﮔر ﺑﺎﺗریھﺎ ﺑﮫ ﺷﮑل ﻧﺎدرﺳت ﻗرار ﮔﯾرﻧد‪ ،‬دﺳﺗﮕﺎه‬‫ﮐﺎر ﻧﺧواھد ﮐرد‪.‬‬‫)ﻋﻼﻣت ‪ (LOW BATTERY‬ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐرد‪ ،‬ھﻣﮫ ﺑﺎﺗریھﺎ را ﺑﺎ ﺑﺎﺗریھﺎی‬‫وﻗﺗﯽ‬‫ﻧو ﺗﻌوﯾض ﮐﻧﯾد‪ .‬ﺑﺎﺗریھﺎی ﮐﮭﻧﮫ و ﻧو را ﺑﺎ ھم در دﺳﺗﮕﺎه ﻗرار ﻧدھﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﻣوﺟب ﮐوﺗﺎه ﺷدن‬‫ﻋﻣر ﺑﺎﺗری ﯾﺎ ﻧﻘص در ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﺑﺎﺗریھﺎ را دو ﯾﺎ ﭼﻧد ﺛﺎﻧﯾﮫ ﺑﻌد از ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه ﺗﻌوﯾض ﮐﻧﯾد‪.‬‬‫اﮔر‬‫)ﻋﻼﻣت ‪ (LOW BATTERY‬ﺣﺗﯽ ﭘس از ﺗﻌوﯾض ﺑﺎﺗریھﺎ ظﺎھر ﺷد‪ ،‬ﯾﮏﺑﺎر اﻧدازهﮔﯾری‬‫ﻓﺷﺎر ﺧون را اﻧﺟﺎم دھﯾد‪ .‬آﻧﮕﺎه دﺳﺗﮕﺎه ﻣﯽﺗواﻧد ﺑﺎﺗریھﺎی ﺟدﯾد را ﺷﻧﺎﺳﺎﯾﯽ ﮐﻧد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﺎﺗریھﺎ ﺧﺎﻟﯽ ﺷده ﺑﺎﺷﻧد‪،‬‬‫)ﻋﻼﻣت ‪ (LOW BATTERY‬ظﺎھر ﻧﻣﯽﺷود‪.‬‬‫ﻋﻣر ﺑﺎﺗری ﺑﺳﺗﮫ ﺑﮫ دﻣﺎی ﭘﯾراﻣوﻧﯽ ﻣﯽﺗواﻧد ﻣﺗﻐﯾر ﺑﺎﺷد و در دﻣﺎی ﮐم ﻣﻣﮑن اﺳت ﻋﻣر ﺑﺎﺗری ﮐﻣﺗر ﺷود‪.‬‬‫ﻣﻌﻣوﻻً‪ ،‬ﭼﮭﺎر ﺑﺎﺗری ﻧوی ‪ LR6‬در دﺳﺗﮕﺎھﯽ ﮐﮫ دو ﺑﺎر در روز اﺳﺗﻔﺎده ﻣﯽﺷود‪ ،‬ﺣدودا ً ﯾﮏ ﺳﺎل ﻋﻣر‬‫ﻣﯽﮐﻧد‪.‬‬‫ﺻرﻓﺎ ً از ﺑﺎﺗریھﺎی ﺗﻌﯾﯾنﺷده اﺳﺗﻔﺎده ﮐﻧﯾد‪ .‬ﺑﺎﺗریھﺎﯾﯽ ﮐﮫ ھﻣراه ﺑﺎ دﺳﺗﮕﺎه ﺑﮫ ﺷﻣﺎ داده ﻣﯽﺷوﻧد‪ ،‬ﺑرای‬‫آزﻣﺎﯾش ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ھﺳﺗﻧد و ﻟذا ﻣﻣﮑن اﺳت ﻋﻣر ﻣﺣدودی داﺷﺗﮫ ﺑﺎﺷﻧد‪.‬‬‫اﮔر ﻗرار اﺳت ﮐﮫ از دﺳﺗﮕﺎه ﺑرای ﻣدت طوﻻﻧﯽ اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪ ،‬ﺑﺎﺗریھﺎﯾش را ﺧﺎرج ﮐﻧﯾد‪.‬‬‫ﻣﻣﮑن اﺳت ﺑﺎﺗری ﻧﺷت ﮐﻧد و ﺑﺎﻋث اﺧﺗﻼل در ﮐﺎرﮐرد دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪9‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻣﺗﺻل ﮐردن ﺷﯾﻠﻧﮓ ھوا‬‫ﻧری راﺑط ھوا را ﺑﮫ داﺧل ﻣﺎدﮔﯽ ھوا وارد ﮐرده‬‫و از ﻣﺣﮑم ﺑودن آن اطﻣﯾﻧﺎن ﺣﺎﺻل ﮐﻧﯾد‪.‬‬‫ﻣﺎدﮔﯽ ھوا‬‫ﻧری راﺑط ھوا‬‫ﻣﺗﺻل ﮐردن آداﭘﺗور ‪AC‬‬‫ﻧری آداﭘﺗور ‪ AC‬را ﺑﮫ ﻓﯾش ‪ DC‬وارد ﮐﻧﯾد‪.‬‬‫ﺳﭘس‪ ،‬آداﭘﺗور ‪ AC‬را ﺑﮫ ﭘرﯾز ﺑرق ﻣﺗﺻل ﮐﻧﯾد‪.‬‬‫از آداﭘﺗور ‪ AC‬ﻣﺷﺧصﺷده اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫)ﺑﮫ ﺻﻔﺣﮫ ‪ 22‬ﻣراﺟﻌﮫ ﮐﻧﯾد‪(.‬‬‫ﻓﯾش ‪DC‬‬‫ﻧری آداﭘﺗور ‪AC‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد آداﭘﺗور ‪ AC‬را از ﭘرﯾز ﺑرق ﺟدا ﮐﻧﯾد‪ ،‬ﺑدﻧﮫ آداﭘﺗور ‪ AC‬را ﺑﮕﯾرﯾد و ﺑﮑﺷﯾد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد ﻓﯾش آداﭘﺗور ‪ AC‬را از دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺟدا ﮐﻧﯾد‪ ،‬ﻓﯾش آداﭘﺗور ‪AC‬‬‫را ﮔرﻓﺗﮫ و از دﺳﺗﮕﺎه ﺑﯾرون ﺑﮑﺷﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪10‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫اﻧﺗﺧﺎب ﺳﺎﯾز ﻣﻧﺎﺳب ﺑﺎزوﺑﻧد‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧدی ﺑﺎ اﻧدازه ﻣﻧﺎﺳب ﻧﻘش ﻣﮭﻣﯽ در ﻗراﺋت ﺻﺣﯾﺢ دﺳﺗﮕﺎه دارد‪ .‬اﮔر اﻧدازه ﺑﺎزوﺑﻧد ﻣﻧﺎﺳب‬‫ﻧﺑﺎﺷد‪ ،‬ﻣﻣﮑن اﺳت ﻗراﺋت دﺳﺗﮕﺎه دﭼﺎر اﺷﺗﺑﺎه ﺷود‪.‬‬‫اﻧدازه ﺑﺎزو روی ھر ﺑﺎزوﺑﻧد ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺷﺎﺧص و داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب روی ﺑﺎزوﺑﻧد ﺑﮫ ﺷﻣﺎ ﻣﯽﮔوﯾد ﮐﮫ آﯾﺎ از ﺑﺎزوﺑﻧد ﺻﺣﯾﺣﯽ اﺳﺗﻔﺎده ﻣﯽﮐﻧﯾد‬‫ﯾﺎ ﺧﯾر‪) .‬ﺑﮫ ﺑﺧش "ﻧﻣﺎدھﺎی ﭼﺎپﺷده روی ﺑﺎزوﺑﻧد" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد‪(.‬‬‫اﮔر ﺷﺎﺧص ﺧﺎرج از داﻣﻧﮫ ﻣورد ﻧظر اﺳت‪ ،‬ﺟﮭت ﺧرﯾد ﺑﺎزوﺑﻧد دﯾﮕر ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽﺗﺎن ﺗﻣﺎس‬‫ﺑﮕﯾرﯾد‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫﻋﻧوان ﯾﮏ ﮐﺎﻻی ﻣﺻرﻓﯽ ﻗﻠﻣداد ﻣﯽﺷود‪ .‬اﮔر ﺑﺎزوﺑﻧد ﭘﺎره ﺷد‪ ،‬ﯾﮑﯽ دﯾﮕر ﺑﺧرﯾد‪.‬‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫اﻧدازه ﺗوﺻﯾﮫ ﺷده ﺑﺎزوﺑﻧد‬‫اﻧدازه ﺑﺎزو‬‫‪ 31‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 45‬ﺳﺎﻧﺗﯽﻣﺗر ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 42‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 32‬ﺳﺎﻧﺗﯽﻣﺗر‬‫اﻧدازه ﺑﺎزو‪ :‬ﻣﺣﯾط ﺑﺎزو در ﺑﺧش ﻋﺿﻠﮫ دو ﺳر‪.‬‬‫‪ UA-767S‬ﺑرای اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد ﮐوﭼﮏ طراﺣﯽ ﻧﺷده اﺳت‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد‬‫‪ .1‬ھﻣﺎنطور ﮐﮫ در ﺗﺻوﯾر ﻧﺷﺎن داده ﺷده‪ ،‬ﺑﺎزوﺑﻧد را‬‫دور ﺑﺎزو‪ ،‬ﯾﻌﻧﯽ ‪ 1‬ﺗﺎ ‪ 2‬ﺳﺎﻧﺗﯽﻣﺗر ﺑﺎﻻﺗر از ﮔودی‬‫آرﻧﺞ ﺑﺑﻧدﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را ﻣﺳﺗﻘﯾﻣﺎ ً روی ﭘوﺳت ﻗرار دھﯾد‪ ،‬زﯾرا‬‫ﻟﺑﺎس ﻣﯽﺗواﻧد ﺑﺎﻋث ﺗﺿﻌﯾف ﺿرﺑﺎن ﺷود و در‬‫ﻗراﺋت ﺧطﺎ اﯾﺟﺎد ﮐﻧد‪.‬‬‫‪ .2‬ﺗﻧﮓ ﺷدن ﺑﺎزو ﺑر اﺛر ﺑﺎﻻ زدن زﯾﺎد آﺳﺗﯾن‬‫ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺧطﺎ در ﻗراﺋت دﺳﺗﮕﺎه ﺷود‪.‬‬‫‪ .3‬ﻣطﻣﺋن ﺷوﯾد ﮐﮫ ﺷﺎﺧص‬‫ﻣﻧﺎﺳﺑﯽ ﻗرار دارد‪.‬‬‫در داﻣﻧﮫ اﻧدازه‬‫آﺳﺗﯾنھﺎ را زﯾﺎد ﺑﺎﻻ ﻧزﻧﯾد‪.‬‬‫‪ 2 - 1‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺷﯾﻠﻧﮓ ھوا‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﭼﺳب ﺑﺎزوﺑﻧد‬‫را ﻣﺣﮑم ﮐﻧﯾد‬‫ﺷﺎﺧص‬‫ﻓﺎرﺳﯽ ‪11‬‬‫ﺑﺎزوﺑﻧد‬‫و داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﻣﺎدھﺎی ﭼﺎپ ﺷده روی ﺑﺎزوﺑﻧد‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﺷﺎﺧص‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‪.‬‬‫اﯾن ﻋﻼﻣت روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﭼﺎپ ﺷده اﺳت‪.‬‬‫▲‬‫‪REF‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫را روی ﺷرﯾﺎن ﺑﺎزو ﯾﺎ‬‫ﻧﺷﺎﻧﮫ‬‫ھمراﺳﺗﺎ ﺑﺎ اﻧﮕﺷت ﺣﻠﻘﮫ در داﺧل‬‫ﺑﺎزو ﺗﻧظﯾم ﮐﻧﯾد‪.‬‬‫از ﺑﺎزوﺑﻧد ﺑزرگ ﺑزرﮔﺳﺎﻻن ﺑﮫﺟﺎی‬‫ﺧﺎرج از داﻣﻧﮫای ﮐﮫ روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎل و‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﯾﺎ ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﻋرﯾض ﭼﺎپ ﻣﯽﺷود‪.‬‬‫ﻋرﯾض اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﻋرﯾض‪ .‬روی‬‫ﺑﺎزوﺑﻧد ﻋرﯾض ﭼﺎپ ﺷده اﺳت‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ‬‫ﺑزرﮔﺳﺎﻻن‪ .‬اﯾن ﻋﻼﻣت روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ‬‫ﺑزرﮔﺳﺎل ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺧﺎرج از داﻣﻧﮫای ﮐﮫ روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎل و‬‫ﺑﺎزوﺑﻧد ﻋرﯾض ﭼﺎپ ﻣﯽﺷود‪.‬‬‫ﮐﻣﺗر از داﻣﻧﮫای روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ از ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﺑﮫﺟﺎی ﺑﺎزوﺑﻧد‬‫ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫ﺑزرﮔﺳﺎﻻن‪.‬‬‫ﺷﻣﺎره دﺳﺗﮫ‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫‪L‬‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن‬‫‪A‬‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬

‫‪L‬‬

‫‪W‬‬

‫‪L‬‬

‫‪A‬‬

‫ﻓﺎرﺳﯽ ‪12‬‬‫‪S‬‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪S‬‬

‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﺣوه اﻧﺟﺎم اﻧدازهﮔﯾری دﻗﯾﻖ‬‫ﺑﮫﻣﻧظور اﻧﺟﺎم دﻗﯾﻖﺗرﯾن اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪:‬‬‫آرام روی ﺻﻧدﻟﯽ ﺑﻧﺷﯾﻧﯾد‪ .‬دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﺑﮕذارﯾد‪ .‬ﭘﺎھﺎﯾﺗﺎن را رویھم ﻧﯾﻧدازﯾد‪ .‬ﭘﺎھﺎﯾﺗﺎن را‬‫ﺻﺎف روی زﻣﯾن ﺑﮕذارﯾد و ﮐﻣرﺗﺎن را ﺻﺎف ﮐﻧﯾد‪.‬‬‫ﺑﮫ ﻣدت ﭘﻧﺞ اﻟﯽ ده دﻗﯾﻘﮫ آرام ﺑﺎﺷﯾد و ﺳﭘس اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫ﻣرﮐز ﺑﺎزوﺑﻧد را ھمﺗراز ﺑﺎ ﻗﻠﺑﺗﺎن ﻗرار دھﯾد‪.‬‬‫در طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت و آرام ﺑﻣﺎﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻧﺑﺎﯾد ﺑﻼﻓﺎﺻﻠﮫ ﭘس از ورزش ﯾﺎ ﺣﻣﺎم ﮐردن اﻧﺟﺎم ﺷود‪ .‬ﺑﯾﺳت اﻟﯽ ﺳﯽ دﻗﯾﻘﮫ آرام ﺑﻣﺎﻧﯾد و ﺑﻌد‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫ﺳﻌﯽ ﮐﻧﯾد اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون را ھر روز در ﯾﮏ ﺳﺎﻋت ﻣﻌﯾن اﻧﺟﺎم دھﯾد‪.‬‬‫اﻧدازهﮔﯾری‬‫در طول اﻧدازهﮔﯾری‪ ،‬طﺑﯾﻌﯽ اﺳت ﮐﮫ اﺣﺳﺎس ﮐﻧﯾد ﺑﺎزوﺑﻧد ﺧﯾﻠﯽ ﻣﺣﮑم ﺑﺳﺗﮫ ﺷده اﺳت‪) .‬ﻧﮕران ﻧﺑﺎﺷﯾد‪(.‬‬‫ﭘس از اﻧدازهﮔﯾری‬‫ﭘس از اﻧدازهﮔﯾری‪ ،‬دﮐﻣﮫ ‪ START‬را ﺑزﻧﯾد ﺗﺎ دﺳﺗﮕﺎه ﺧﺎﻣوش ﺷود‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑﺎز ﮐﻧﯾد و اطﻼﻋﺎت را ﺛﺑت ﮐﻧﯾد‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﯾن دﺳﺗﮕﺎه دارای ﻋﻣﻠﮑرد ﺧﺎﻣوش ﮐردن ﺧودﮐﺎر اﺳت ﮐﮫ ﺣدود ﯾﮏ دﻗﯾﻘﮫ ﭘس از اﻧدازهﮔﯾری‬‫دﺳﺗﮕﺎه را ﺧﺎﻣوش ﻣﯽﮐﻧد‪.‬‬‫در ﺻورت ﻧﯾﺎز ﺑﮫ ﺗﮑرار اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد‪ ،‬ﺣداﻗل ﺳﮫ دﻗﯾﻘﮫ ﻣﯾﺎن ھر ﺑﺎر اﻧدازهﮔﯾری‬‫ﻓﺎﺻﻠﮫ ﺑﯾﻧدازﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪13‬‬‫اﻧدازهﮔﯾریھﺎ‬‫ﭘﯾش از اﻧدازهﮔﯾری‪ ،‬ﺑﮫ "ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻋﺎدی‬‫‪.1‬‬‫ﺑﺎزوﺑﻧد را دور ﺑﺎزو )ﺗرﺟﯾﺣﺎ ً روی ﺑﺎزوی ﭼپ( ﻗرار دھﯾد‪.‬‬‫در ﺣﯾن اﻧدازهﮔﯾری آرام ﺑﻣﺎﻧﯾد‪.‬‬‫‪.2‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ھﻣﮫ ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر ﻧﻣﺎﯾﺎن ﻣﯽﺷوﻧد‪.‬‬‫ﺻﻔر )‪ (0‬ﻧﻣﺎﯾﺎن ﻣﯽﺷود ﮐﮫ اﻧدﮐﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ اﻧدازهﮔﯾری آﻏﺎز ﻣﯽﺷود‪ ،‬ھﻣﺎنطور ﮐﮫ در‬‫ﺗﺻوﯾر ﺳﻣت راﺳت ﻧﺷﺎن داده ﺷده اﺳت‪ ،‬ﻧﻣﺎﯾﺷﮕر ﺗﻐﯾﯾر‬‫ﻣﯽﮐﻧد‪ .‬ﺑﺎزوﺑﻧد ﺷروع ﺑﮫ ﺑﺎد ﺷدن ﻣﯽﮐﻧد‪ .‬اﺣﺳﺎس ﺗﻧﮕﯽ‬‫ﺑﺎزوﺑﻧد طﺑﯾﻌﯽ اﺳت‪ .‬ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر در ﻟﺑﮫ ﺳﻣت ﭼپ‬‫ﻧﻣﺎﯾﺷﮕر در طول ﺑﺎد ﮐردن ﻧﻣﺎﯾﺎن ﻣﯽﺷود‪.‬‬‫ﻧﮑﺗﮫ‪ :‬ھر زﻣﺎن ﺧواﺳﺗﯾد ﺑﺎد ﮐردن را ﻣﺗوﻗف ﮐﻧﯾد‪ ،‬دﮐﻣﮫ‬‫‪ START‬را ﻣﺟددا ً ﻓﺷﺎر دھﯾد‪.‬‬‫‪.3‬‬‫وﻗﺗﯽ ﺑﺎد ﮐردن ﺗﻣﺎم ﺷد‪ ،‬ﺧﺎرج ﺷدن ﺑﺎد ﺑﮫﺻورت ﺧودﮐﺎر‬‫آﻏﺎز ﺷده و‬‫)ﻋﻼﻣت ﻗﻠب( ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن‬‫ﻣﯽﮐﻧد‪ ،‬ﮐﮫ ﻧﺷﺎن ﻣﯽدھد اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺿرﺑﺎن ﻗﻠب ﺗﺷﺧﯾص داده ﺷد‪ ،‬ﻋﻼﻣت ﻗﻠب ﺑﺎ ھر‬‫ﺿرﺑﺎن ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫‪START‬‬‫را ﻓﺷﺎر دھﯾد‬‫ﮐﻠﯾﮫ ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر‬‫ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‬‫ﻧﻣﺎﯾش ﺻﻔر‬‫ﺷروع ﺑﺎد ﺷدن‬‫در ﺣﺎل اﻓزاﯾش ﻓﺷﺎر‬‫اﮔر ﻓﺷﺎر ﻣﻧﺎﺳب ﺑﮫ دﺳت ﻧﯾﺎﻣد‪ ،‬دﺳﺗﮕﺎه ﺑﮫطور‬‫ﺧودﮐﺎر ﻣﺟددا ً ﺷروع ﺑﮫ ﺑﺎد ﺷدن ﻣﯽﮐﻧد‪.‬‬‫ﺑرای اﺟﺗﻧﺎب از ﺑﺎد ﺷدن ﻣﺟدد‪ ،‬ﺑﮫ ﺑﺧش‬‫"اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب" در‬‫ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫‪.4‬‬‫وﻗﺗﯽ اﻧدازهﮔﯾری ﺗﻣﺎم ﺷد‪ ،‬ﻗراﺋت ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ و دﯾﺎﺳﺗوﻟﯽ‬‫و ﺿرﺑﺎن ﻧﺑض ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‪.‬‬‫ھوای ﺑﺎﻗﯽﻣﺎﻧده از ﺑﺎزوﺑﻧد ﺧﺎرج ﺷده و ﺑﺎد آن ﺑﮫطور ﮐﺎﻣل‬‫ﺧﺎﻟﯽ ﻣﯽﺷود‪.‬‬‫‪.5‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد ﺗﺎ دﺳﺗﮕﺎه ﺧﺎﻣوش ﺷود‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد را ﺑﺎ ﻓﺎﺻﻠﮫ ﺣداﻗل‬‫ﺳﮫ دﻗﯾﻘﮫ اﻧﺟﺎم دھﯾد‪.‬‬‫اﻧدازهﮔﯾری در ﺣﺎل‬‫اﻧﺟﺎم اﺳت‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫طﺑﻘﮫﺑﻧدی ‪WHO‬‬‫ﺿرﺑﺎن ﻧﺑض‬‫ھوای ﺑﺎﻗﯽﻣﺎﻧده را ﺑﮫﺻورت‬‫ﺧودﮐﺎر ﺑﯾرون ﻣﯽدھد‬‫ﻓﺎرﺳﯽ ‪14‬‬‫اﻧدازهﮔﯾریھﺎ‬‫اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫‪ UA-767S‬ﺑرای ﺷﻧﺎﺳﺎﯾﯽ ﺿرﺑﺎن ﻗﻠب و ﺑﺎد ﮐردن ﺧودﮐﺎر ﺑﺎزوﺑﻧد ﺗﺎ رﺳﯾدن ﺑﮫ ﺳطﺢ ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ ﻣﺷﺧص طراﺣﯽ ﺷده اﺳت‪.‬‬‫از اﯾن روش ھﻧﮕﺎﻣﯽ اﺳﺗﻔﺎده ﮐﻧﯾد ﮐﮫ ﺑﺎد ﮐردن ﻣﺟدد ﭼﻧدﯾن ﺑﺎر ﺗﮑرار ﻣﯽﺷود ﯾﺎ ھﻧﮕﺎﻣﯽ ﮐﮫ ﺣﺗﯽ ﺑﺎ‬‫ﮐﺎھش ﻓﺷﺎر ﺑﮫ ‪ mmHg 20‬ﯾﺎ ﮐﻣﺗر‪ ،‬ﻧﺗﺎﯾﺞ ﻧﺷﺎن داده ﻧﻣﯽﺷوﻧد‪.‬‬‫‪.1‬‬‫ﺑﺎزوﺑﻧد را دور ﺑﺎزو )ﺗرﺟﯾﺣﺎ ً روی ﺑﺎزوی ﭼپ( و ھمﺗراز ﺑﺎ‬‫ﻗﻠب ﺑﺑﻧدﯾد‪.‬‬‫‪.2‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫‪.3‬‬‫زﻣﺎﻧﯽ ﮐﮫ ﻋﻼﻣت ﺻﻔر ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐرد‪،‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد و ﻧﮕﮫدارﯾد ﺗﺎ‬‫ﻋددی در ﺣدود ‪ 30‬ﺗﺎ ‪ mmHg 40‬ﺑﺎﻻﺗر از ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ ﻣورد اﻧﺗظﺎر ظﺎھر ﺷود‪.‬‬‫‪.4‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﮫ رﻗم ﻣورد ﻧظر رﺳﯾدﯾد‪ ،‬دﮐﻣﮫ‬‫‪ START‬را رھﺎ ﮐﻧﯾد ﺗﺎ اﻧدازهﮔﯾری آﻏﺎز ﺷود‪ .‬ﺑﮫ‬‫ھﻣﺎن ﺗرﺗﯾﺑﯽ ﮐﮫ در ﺻﻔﺣﮫ ﻗﺑل ﺗوﺿﯾﺢ داده ﺷد‪ ،‬ﺑﮫ‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧوﻧﺗﺎن اداﻣﮫ دھﯾد‪.‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫‪ START‬دﮐﻣﮫ را‬‫ﻓﺷﺎر دھﯾد‬‫دﮐﻣﮫ را در ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻣطﻠوب رھﺎ ﮐﻧﯾد‬‫‪START‬‬‫دﮐﻣﮫ را ﻓﺷﺎر دھﯾد و‬‫ﻧﮕﮫدارﯾد‬‫ﺑرای ﻣﺷﺎھده ﺟزﺋﯾﺎت ﺑﯾﺷﺗر در ﻣورد‬‫اﻧدازهﮔﯾری ﺑﮫ ﺻﻔﺣﮫ ﻗﺑل ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ‬‫در ﻣوﻗﻌﯾت راﺣﺗﯽ ﺑﻧﺷﯾﻧﯾد‪ .‬دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﻗرار دھﯾد ﺑﮫ ﮔوﻧﮫای ﮐﮫ ﮐف دﺳتھﺎﯾﺗﺎن رو ﺑﮫ‬‫ﺑﺎﻻ ﺑوده و ﺑﺎزوﺑﻧد و ﻗﻠﺑﺗﺎن ھمﺗراز ﺑﺎﺷﻧد‪.‬‬‫ﭘﯾش از آﻏﺎز اﻧدازهﮔﯾری‪ ،‬ﺣدود ﭘﻧﺞ ﺗﺎ ده دﻗﯾﻘﮫ آرام ﺑﺎﺷﯾد‪ .‬اﮔر ﺑﮫ دﻟﯾل ﺗﻧشھﺎی ھﯾﺟﺎﻧﯽ دﭼﺎر ھﯾﺟﺎن ﯾﺎ‬‫اﻓﺳردﮔﯽ ھﺳﺗﯾد‪ ،‬ﻓﺷﺎر ﺧوﻧﺗﺎن ﺑﺎﻻﺗر )ﯾﺎ ﭘﺎﯾﯾنﺗر( از ﻓﺷﺎر ﺧون ﻧرﻣﺎل ﻧﺷﺎن داده ﺧواھد ﺷد‪ ،‬و ﺿرﺑﺎن‬‫ﻗﻠب ﻧﯾز ﻣﻌﻣوﻻً ﺳرﯾﻊﺗر از ﺣﺎﻟت ﻧرﻣﺎل ﻣﯽﺑﺎﺷد‪.‬‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد ﻣرﺗﺑﺎ ً در ﺣﺎل ﺗﻐﯾﯾر اﺳت و ﺑﮫ ﮐﺎری ﮐﮫ اﻧﺟﺎم ﻣﯽدھد و ﻏذاﯾﯽ ﮐﮫ ﺧورده اﺳت‬‫ﺑﺳﺗﮕﯽ دارد‪ .‬آﻧﭼﮫ ﻧوﺷﯾدهاﯾد ﻣﯽﺗواﻧد ﺗﺄﺛﯾر ﺑﺳﯾﺎر ﻗوی و ﺳرﯾﻌﯽ روی ﻓﺷﺎر ﺧوﻧﺗﺎن داﺷﺗﮫ ﺑﺎﺷد‪.‬‬‫ﻣﺑﻧﺎی اﻧدازهﮔﯾریھﺎی اﯾن دﺳﺗﮕﺎه ﺿرﺑﺎن ﻗﻠب ﻣﯽﺑﺎﺷد‪ .‬اﮔر ﺿرﺑﺎن ﻗﻠب ﺑﺳﯾﺎر ﺿﻌﯾف ﯾﺎ ﻧﺎﻣﻧظﻣﯽ‬‫داﺷﺗﮫ ﺑﺎﺷﯾد‪ ،‬ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫﺳﺧﺗﯽ ﺑﺗواﻧد ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﺷﺧﯾص دھد‪.‬‬‫اﮔر دﺳﺗﮕﺎه وﺟود ﺷراﯾط ﻏﯾر طﺑﯾﻌﯽ را ﺗﺷﺧﯾص دھد‪ ،‬اﻧدازهﮔﯾری را ﻣﺗوﻗف ﻣﯽﮐﻧد و ﻧﻣﺎد ﺧطﺎ ظﺎھر‬‫ﻣﯽﺷود‪ .‬ﺑرای ﻣﺷﺎھده ﺗوﺿﯾﺢ ﻧﻣﺎدھﺎ‪ ،‬ﺑﮫ ﺻﻔﺣﮫ ‪ 7‬ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪15‬‬‫اﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺑرای اﺳﺗﻔﺎده ﺑزرﮔﺳﺎﻻن طراﺣﯽ ﺷده اﺳت‪ .‬اﮔر ﻣﯽﺧواھﯾد از اﯾن دﺳﺗﮕﺎه‬‫ﺑرای ﮐودﮐﺎن اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪ .‬ﮐودﮐﺎن ﻧﺑﺎﯾد ﺑدون ﮐﻣﮏ ﺑزرﮔﺳﺎﻻن از اﯾن‬‫دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﻧد‪.‬‬‫ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ﺧودﮐﺎر ﭘﺎﯾش ﻓﺷﺎر ﺧون ﻣﯽﺗواﻧد ﺗﺣت ﺗﺄﺛﯾر دﻣﺎ ﯾﺎ رطوﺑت ﺷدﯾد ﯾﺎ ارﺗﻔﺎع ﻗرار ﮔﯾرد‪.‬‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ‬‫ﻧﮑﺗﮫ‪:‬‬‫اﯾن دﺳﺗﮕﺎه ‪ 60‬اﻧدازهﮔﯾری آﺧر را در ﺣﺎﻓظﮫاش ذﺧﯾره ﻣﯽﮐﻧد‪.‬‬‫‪ .1‬ھﻧﮕﺎﻣﯽ ﮐﮫ ھﯾﭻ ﭼﯾزی ﻧﻣﺎﯾش داده ﻧﻣﯽﺷود‪،‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر داده و ﻧﮕﮫدارﯾد ﺗﺎ‬‫اطﻼﻋﺎت ذﺧﯾرهﺷده ﻓراﺧواﻧﯽ ﺷود‪.‬‬‫ﻓﺷﺎر دھﯾد و‬‫ﻧﮕﮫ دارﯾد‬‫‪ START‬دﮐﻣﮫ‬‫ﻣﯾﺎﻧﮕﯾن ﺳﯾﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن دﯾﺎﺳﺗوﻟﯽ‬‫‪ .2‬ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت ﻧﺷﺎن داده ﺷد‪ ،‬دﮐﻣﮫ‬‫را رھﺎ ﮐﻧﯾد‪.‬‬‫ﻣﯾﺎﻧﮕﯾن ﺿرﺑﺎن‬‫ﻣﯾﺎﻧﮕﯾن‬‫ﺗﻣﺎم اطﻼﻋﺎت‬‫‪ .3‬ﺷﻣﺎره اطﻼﻋﺎت و اطﻼﻋﺎت ذﺧﯾره ﺷده ﺑﮫﺻورت‬‫ﺧودﮐﺎر و ﺑﮫ ﺗرﺗﯾب از آﺧرﯾن اﻧدازهﮔﯾری ﻧﺷﺎن‬‫داده ﻣﯽﺷوﻧد‪.‬‬‫ﺳﯾﺳﺗوﻟﯽ‬‫‪ .4‬ﭘس از آن ﮐﮫ ﺗﻣﺎﻣﯽ اطﻼﻋﺎت ﻧﻣﺎﯾش داده ﺷد‪،‬‬‫ﻧﻣﺎﯾﺷﮕر ﺧﺎﻣوش ﻣﯽﺷود‪.‬‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﺿرﺑﺎن‬‫ﺟدﯾدﺗرﯾن داده‬‫ﺗوﺟﮫ‪:‬‬‫اﮔر دﮐﻣﮫ ‪ START‬را در زﻣﺎن ﻓراﺧواﻧﯽ‬‫اطﻼﻋﺎت ﻓﺷﺎر دھﯾد‪ ،‬دﺳﺗﮕﺎه ﺧﺎﻣوش ﻣﯽﺷود‪.‬‬‫ﺳﯾﺳﺗوﻟﯽ‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﺿرﺑﺎن‬‫آﺧرﯾن داده )ﻗدﯾﻣﯽﺗرﯾن(‬‫ﻓﺎرﺳﯽ ‪16‬‬‫ﺷﺎﺧص ‪ IHB/AFib‬ﭼﯾﺳت؟‬‫وﻗﺗﯽ دﺳﺗﮕﺎه رﯾﺗم ﻧﺎﻣﻧظﻣﯽ را در طول اﻧدازهﮔﯾری ﺷﻧﺎﺳﺎﯾﯽ ﮐﻧد‪ ،‬ﻧﺷﺎﻧﮕر ‪ IHB/AFib‬روی ﻧﻣﺎﯾﺷﮕر‬‫ھﻣراه ﺑﺎ ﻣﻘﺎدﯾر اﻧدازهﮔﯾری ظﺎھر ﻣﯽﺷود‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫ﺗوﺻﯾﮫ ﻣﯽﺷود ﮐﮫ در ﺻورت ﻣﺷﺎھده اﯾن‬‫ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫ﻓرﮐﺎﻧس ﺷﺎﺧص ‪ ،IHB/AFib‬ﺑﺎ ﭘزﺷﮑﺗﺎن‬‫‪ AFib‬ﭼﯾﺳت؟‬‫ﻗﻠب ﺑﮫ دﻟﯾل ﺳﯾﮕﻧﺎلھﺎی اﻟﮑﺗرﯾﮑﯽ ﮐﮫ در ﻗﻠب روی ﻣﯽدھد‪ ،‬ﻣﻧﻘﺑض ﻣﯽﺷود و ﺧون را ﺑﮫ ﮐل ﺑدن‬‫ﻣﯽﻓرﺳﺗد‪ .‬ﻓﯾﺑرﯾﻼﺳﯾون دھﻠﯾزی )‪ (AFib‬زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ ﺳﯾﮕﻧﺎل اﻟﮑﺗرﯾﮑﯽ در دھﻠﯾز ﻗﻠب ﻣﻐﺷوش‬‫ﺷده و ﻣﻧﺟر ﺑﮫ آﺷﻔﺗﮕﯽ در ﻓواﺻل ﺿرﺑﺎن ﻣﯽﺷود‪ AFib .‬ﻣﯽﺗواﻧد ﻣوﺟب ﺷود ﮐﮫ ﺧون در ﻗﻠب از‬‫ﺣرﮐت ﺑﺎزاﯾﺳﺗد ﮐﮫ ﺑﮫﺳﺎدﮔﯽ ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﻟﺧﺗﮫ ﺧون و ﺑﮫ ﺗﺑﻊ آن ﻣوﺟب ﺳﮑﺗﮫ و ﺣﻣﻠﮫ ﻗﻠﺑﯽ ﺷود‪.‬‬‫‪％IHB/AFib‬‬‫‪ %IHB/AFib‬ﺑﮫﺻورت ﻣﯾزان ﺗﮑرار ‪ IHB/AFib‬ﺷﻧﺎﺳﺎﯾﯽ ﺷده ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫‪ IHB/AFib‬ﻧﮫ ﺗﻧﮭﺎ ﻣﯽﺗواﻧد ﻧوﯾزھﺎﯾﯽ ﻣﺎﻧﻧد ﺣرﮐت ﻓﯾزﯾﮑﯽ را ﺗﺷﺧﯾص دھد‪ ،‬ﺑﻠﮑﮫ ﻗﺎدر اﺳت ﺿرﺑﺎن‬‫ﻗﻠب ﻧﺎﻣﻧظم را ﻧﯾز ﺷﻧﺎﺳﺎﯾﯽ ﻧﻣﺎﯾد‪ .‬در ﻧﺗﯾﺟﮫ‪ ،‬ﺗوﺻﯾﮫ ﻣﯽﮐﻧﯾم ﮐﮫ اﮔر ﻣﯾزان ‪ %IHB/AFib‬ﺑﺎﻻﺳت‪ ،‬ﺑﺎ‬‫ﭘزﺷﮑﺗﺎن ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫ﺗﻌداد ‪IHB/AFibs‬‬‫ﺷﻧﺎﺳﺎﯾﯽﺷده در ﺣﺎﻓظﮫ‬‫‪[%] 100 x‬‬‫‪= %IHB/AFib‬‬‫ﺗﻌداد ﮐل‬‫ﻧﻣﺎﯾش ‪ %IHB/AFib %IHB/AFib:‬زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ ﻣﻘﺎدﯾر ﻣﯾﺎﻧﮕﯾن ﻧﺷﺎن داده ﻣﯽﺷوﻧد‪.‬‬‫)ﺑﮫ ﺑﺧش "‪ .2‬ﻓراﺧواﻧﯽ اطﻼﻋﺎت" در "ﺣﺎﻟت ﻋﻣﻠﯾﺎت" ﻣراﺟﻌﮫ ﮐﻧﯾد(‬‫‪ %IHB/AFib‬زﻣﺎﻧﯽ ﮐﮫ ﺗﻌداد اطﻼﻋﺎت ﺣﺎﻓظﮫ ﺷش ﻣورد ﯾﺎ ﮐﻣﺗر اﺳت‪،‬‬‫ﻧﻣﺎﯾش داده ﻧﻣﯽﺷود‪.‬‬‫ﻧﻣﺎﯾش ﻣﻘدار ﻣﯾﺎﻧﮕﯾن‬‫ﺳطﺢ ‪0‬‬‫‪0=%IHB/AFib‬‬‫ﺳطﺢ ‪1‬‬‫‪9 - 1=%IHB/AFib‬‬‫ﺳطﺢ ‪3‬‬‫ﺳطﺢ ‪2‬‬‫‪100 - 25=%IHB/AFib 24 - 10=%IHB/AFib‬‬‫ﻋدم ﻧﻣﺎﯾش‬‫ﻓﺎرﺳﯽ ‪17‬‬‫ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر‬‫اﯾن ﻧﺷﺎﻧﮕر ﻣﯾزان ﻓﺷﺎر در طول اﻧدازهﮔﯾری را ﻧﺷﺎن ﻣﯽدھد‪.‬‬‫در ﺣﺎل اﻓزاﯾش ﻓﺷﺎر‬‫آزاد ﺷدن ھوا‬‫ﺑﺎد ﮐردن در ﺣﺎل اﻧﺟﺎم اﺳت‬‫اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‬‫اﺗﻣﺎم ﺑﺎد ﮐردن‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫ھر ﺑﺧش از ﻧوار ﻧﺷﺎﻧﮕر ﻣﺗﻧﺎظر ﺑﺎ ﯾﮏ ﺑﺧش از طﺑﻘﮫﺑﻧدی‬‫ﻓﺷﺎر ﺧون ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪ (WHO‬اﺳت‪.‬‬‫ﻣﺛﺎل‪:‬‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫ﻓﺷﺎر ﺧون ﺷدﯾد‬‫ﻓﺷﺎر ﺧون ﻣﺗوﺳط‬‫ﻓﺷﺎر ﺧون ﺧﻔﯾف‬‫ﻧرﻣﺎل رو ﺑﮫ ﺑﺎﻻ‬‫ﻧرﻣﺎل‬‫ﺑﮭﯾﻧﮫ‬‫ﻓﺷﺎر ﺧون ﻣﺗوﺳط‬‫ﻓﺷﺎر ﺧون ﺧﻔﯾف‬‫‪:‬اﯾن ﺷﺎﺧص ﯾﮏ ﺑﺧش را ﻧﻣﺎﯾش ﻣﯽدھد ﮐﮫ ﺑر ﻣﺑﻧﺎی اطﻼﻋﺎت‬‫ﮐﻧوﻧﯽ و ﻣﺗﻧﺎﺳب ﺑﺎ طﺑﻘﮫﺑﻧدی ‪ WHO‬ﻣﯽﺑﺎﺷد‪.‬‬‫ﻧرﻣﺎل رو ﺑﮫ ﺑﺎﻻ‬‫در ﻣورد ﻓﺷﺎر ﺧون‬‫ﻓﺷﺎر ﺧون ﭼﯾﺳت؟‬‫ﻓﺷﺎر ﺧون ﻧﯾروﯾﯽ اﺳت ﮐﮫ ﺧون ﺑﮫ دﯾواره رگھﺎ وارد ﻣﯽﮐﻧد‪ .‬ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ‬‫ﻗﻠب ﻣﻧﻘﺑض ﻣﯽﺷود‪ .‬ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ ﻗﻠب ﻣﻧﺑﺳط ﻣﯽﺷود‪ .‬ﻓﺷﺎر ﺧون ﺑر ﺣﺳب‬‫ﻣﯾﻠﯽﻣﺗر ﺟﯾوه )‪ (mmHg‬اﻧدازهﮔﯾری ﻣﯽﺷود‪ .‬ﻓﺷﺎر ﺧون طﺑﯾﻌﯽ ﯾﮏ ﻓرد ﺗوﺳط ﻓﺷﺎر ﭘﺎﯾﮫ ﻧﺷﺎن داده‬‫ﻣﯽﺷود ﮐﮫ اول ﺻﺑﺢ زﻣﺎﻧﯽ ﮐﮫ ﻓرد ھﻧوز در ﺣﺎل اﺳﺗراﺣت اﺳت و ﻏذاﯾﯽ ﻧﺧورده اﻧدازهﮔﯾری‬‫ﻣﯽﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪18‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﭼﯾﺳت و ﭼﮕوﻧﮫ ﮐﻧﺗرل ﻣﯽﺷود؟‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﮐﮫ ﺑﮫ ﻣﻌﻧﺎی ﻓﺷﺎر ﺧون ﻏﯾر ﻧرﻣﺎل ﺑﺎﻻی ﺷرﯾﺎﻧﯽ اﺳت‪ ،‬درﺻورﺗﯽﮐﮫ درﻣﺎن ﻧﺷود‬‫ﻣﯽﺗواﻧد ﻣﺷﮑﻼت زﯾﺎدی را ﺑرای ﻗﻠب از ﺟﻣﻠﮫ ﺳﮑﺗﮫ و ﺣﻣﻠﮫ ﻗﻠﺑﯽ اﯾﺟﺎد ﮐﻧد‪ .‬ﻓﺷﺎر ﺧون ﺑﺎﻻ ﻣﯽﺗواﻧد از‬‫طرﯾﻖ ﺗﻐﯾﯾر ﺳﺑﮏ زﻧدﮔﯽ‪ ،‬ﭘرھﯾز از اﺳﺗرس و ﻣﺻرف دارو ﺗﺣت ﻧظر ﭘزﺷﮏ ﮐﻧﺗرل ﺷود‪.‬‬‫ﺑرای ﭘﯾﺷﮕﯾری از اﺑﺗﻼ ﺑﮫ اﻓزاﯾش ﻓﺷﺎر ﺧون‪ ،‬ﺑﺎﯾد آن را ﺗﺣت ﮐﻧﺗرل داﺷﺗﮫ ﺑﺎﺷﯾد‪:‬‬‫ﻣرﺗب ورزش ﮐﻧﯾد‬‫ﺳﯾﮕﺎر ﻧﮑﺷﯾد‬‫ﺑﮫطور ﻣرﺗب ﺑرای ﭼﮏ آپ و ﻣﻌﺎﯾﻧﮫ ﭘزﺷﮑﯽ‬‫ﻣﺻرف ﻧﻣﮏ و ﭼرﺑﯽ را ﮐﺎھش دھﯾد‬‫ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫وزن ﻣﻧﺎﺳب داﺷﺗﮫ ﺑﺎﺷﯾد‬‫ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ اﻧدازهﮔﯾری ﮐﻧم؟‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون در ﮐﻠﯾﻧﯾﮏ ﯾﺎ ﻣطب ﭘزﺷﮏ ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺗرس در ﻓرد ﺷده و اﯾن اﻣر‬‫ﻣﯽﺗواﻧد ﺑﺎﻋث اﻓزاﯾش ﻣﻘدار ﻗراﺋت ﺷده ﺗوﺳط دﺳﺗﮕﺎه ﺑﮫ ﻣﯾزان ‪ 25‬ﺗﺎ ‪ mmHg 30‬ﺑﯾش از ﻓﺷﺎر‬‫ﺧون اﻧدازهﮔﯾری ﺷده در ﻣﻧزل ﺷود‪ .‬اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون در ﺧﺎﻧﮫ ﻣوﺟب ﮐﺎھش ﺗﺄﺛﯾر ﻋواﻣل‬‫ﺧﺎرﺟﯽ ﺑر ﻗراﺋت ﻓﺷﺎر ﺧون ﻣﯽﺷود‪ ،‬ﻣﮑﻣل ﻗراﺋت ﭘزﺷﮏ اﺳت و ﺳﺎﺑﻘﮫ دﻗﯾﻖﺗر و ﮐﺎﻣلﺗری از ﻓﺷﺎر‬‫ﺧوﻧﺗﺎن را در اﺧﺗﯾﺎر ﭘزﺷﮏ ﻗرار ﻣﯽدھد‪.‬‬‫طﺑﻘﮫﺑﻧدی ﻓﺷﺎر ﺧون ‪WHO‬‬‫اﺳﺗﺎﻧداردھﺎی ارزﯾﺎﺑﯽ ﻓﺷﺎر ﺧون ﺑﺎﻻ‪،‬‬‫ﺑدون در ﻧظر ﮔرﻓﺗن ﺳن‪ ،‬ﺗوﺳط‬‫ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫طﺑﻖ ﻧﻣودار‪ ،‬اراﺋﮫ ﺷدهاﻧد‪.‬‬‫ﻧوﺳﺎﻧﺎت ﻓﺷﺎر ﺧون‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد در طول روز و‬‫در ﻓﺻول ﻣﺧﺗﻠف ﺑﮫ ﺷدت ﺗﻐﯾﯾر‬‫ﻣﯽﮐﻧد‪ .‬ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد ﻣﯽﺗواﻧد‬‫ﺑر اﺛر ﺷراﯾط ﻣﺧﺗﻠف در طول روز ﺑﮫ‬‫ﻣﯾزان ‪ 30‬ﺗﺎ ‪ mmHg 50‬ﺗﻐﯾﯾر ﮐﻧد‪ .‬در اﻓرادی ﮐﮫ دﭼﺎر ﻓﺷﺎر ﺧون ﺑﺎﻻ ھﺳﺗﻧد‪ ،‬اﯾن ﻧوﺳﺎﻧﺎت ﺣﺗﯽ‬‫ﺑﯾﺷﺗر ﻧﯾز ﻣﯽﺑﺎﺷد‪ .‬ﻣﻌﻣوﻻً ﻓﺷﺎر ﺧون در ﻣﺣل ﮐﺎر ﯾﺎ در زﻣﺎن ﺑﺎزی اﻓزاﯾش ﯾﺎﻓﺗﮫ و در طول ﺧواب ﺑﮫ‬‫ﮐﻣﺗرﯾن ﻣﯾزان ﺧود ﻣﯽرﺳد‪ .‬ﻟذا ﺑﺎ ﻣﺷﺎھده ﻧﺗﺎﯾﺞ ﯾﮏﺑﺎر اﻧدازهﮔﯾری ﺑﯾش از ﺣد ﻧﮕران ﻧﺷوﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون را ھر روز در زﻣﺎن ﻣﺷﺧﺻﯽ اﻧﺟﺎم داده و از دﺳﺗوراﻟﻌﻣﻠﯽ ﮐﮫ در اﯾن راھﻧﻣﺎ‬‫ﺗوﺿﯾﺢ داده ﺷده اﺳت اﺳﺗﻔﺎده ﮐﻧﯾد ﺗﺎ ﻓﺷﺎر ﺧون‬‫ﻧرﻣﺎﻟﺗﺎن را ﭘﯾدا ﮐﻧﯾد‪ .‬ﻗراﺋت ﻣﻧظم ﻣﯽﺗواﻧد ﺳﺎﺑﻘﮫ‬‫ﺟﺎﻣﻌﯽ از ﻓﺷﺎر ﺧوﻧﺗﺎن را در اﺧﺗﯾﺎرﺗﺎن ﻗرار دھد‪.‬‬‫ھﻧﮕﺎم ﺛﺑت ﻓﺷﺎر ﺧوﻧﺗﺎن‪ ،‬ﺣﺗﻣﺎ ً ﺗﺎرﯾﺦ و ﺳﺎﻋت را ﻗﯾد‬‫ﮐﻧﯾد‪ .‬ﺑرای ﺗﻔﺳﯾر اطﻼﻋﺎت ﻓﺷﺎر ﺧوﻧﺗﺎن‪ ،‬ﺑﺎ ﭘزﺷﮏ‬‫ﻣﺷورت ﮐﻧﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪19‬‬‫ﻋﯾبﯾﺎﺑﯽ‬‫ﻣﺷﮑل‬‫دﻟﯾل اﺣﺗﻣﺎﻟﯽ‬‫ھﯾﭻ ﭼﯾزی ﻧﻣﺎﯾش داده ﺑﺎﺗریھﺎ ﺧﺎﻟﯽ ﺷدهاﻧد‪.‬‬‫ﻧﻣﯽﺷود‪ ،‬ﺣﺗﯽ زﻣﺎﻧﯽ‬‫ﮐﮫ دﺳﺗﮕﺎه روﺷن‬‫اﻧﺗﮭﺎی ﺑﺎﺗریھﺎ در ﻣوﻗﻌﯾت‬‫اﺳت‪.‬‬‫ﺻﺣﯾﺢ ﻗرار ﻧﮕرﻓﺗﮫاﻧد‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﺎد ﻧﻣﯽﺷود‪.‬‬‫وﻟﺗﺎژ ﺑﺎﺗری ﺧﯾﻠﯽ ﮐم اﺳت‪.‬‬‫)ﻋﻼﻣت ‪LOW‬‬‫‪ (BATTERY‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫اﮔر ﺑﺎﺗریھﺎ ﺑﮫطور ﮐﺎﻣل ﺧﺎﻟﯽ‬‫ﺷده ﺑﺎﺷﻧد‪ ،‬اﯾن ﻋﻼﻣت ظﺎھر‬‫ﻧﻣﯽﺷود‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫدرﺳﺗﯽ روی ﺑﺎزو ﯾﺎ ﻣﭻ‬‫ﻗرار داده ﻧﺷده اﺳت‪.‬‬‫در ﺣﯾن اﻧدازهﮔﯾری ﺑﺎزو ﯾﺎ ﺑدﻧﺗﺎن‬‫را ﺗﮑﺎن دادهاﯾد‪.‬‬‫دﺳﺗﮕﺎه اﻧدازهﮔﯾری‬‫ﻧﻣﯽﮐﻧد‪.‬‬‫ﻣﻘﺎدﯾر ﻗراﺋتﺷده ﺧﯾﻠﯽ ﺑﺎزوﺑﻧد در ﻣوﻗﻌﯾت ﺻﺣﯾﺢ ﻗرار‬‫ﺑﺎﻻ ﯾﺎ ﺧﯾﻠﯽ ﭘﺎﯾﯾن‬‫ﻧدارد‪.‬‬‫ھﺳﺗﻧد‪.‬‬‫ﺳﺎﯾر ﻣوارد‬‫ﺗوﺟﮫ‪:‬‬‫ﻋدد ﻧﺷﺎن داده ﺷده ﺗوﺳط دﺳﺗﮕﺎه‬‫ﻣﺗﻔﺎوت از ﻋددی اﺳت ﮐﮫ در‬‫ﮐﻠﯾﻧﯾﮏ ﯾﺎ ﻣطب ﭘزﺷﮏ‬‫اﻧدازهﮔﯾری ﺷده اﺳت‪.‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫ھﻣﮫ ﺑﺎﺗریھﺎ را ﺑﺎ ﺑﺎﺗریھﺎی ﻧو ﺟﺎﯾﮕزﯾن‬‫ﮐﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ را ﺧﺎرج ﮐﻧﯾد و آﻧﮭﺎ را ﻣطﺎﺑﻖ ﺑﺎ‬‫ﻋﻼﻣت ﻣﺛﺑت و ﻣﻧﻔﯽ ﮐﮫ در ﻣﺣﻔظﮫ ﺑﺎطری‬‫ﻧوﺷﺗﮫ ﺷده اﺳت‪ ،‬در ﺟﺎی ﺧود ﻗرار دھﯾد‪.‬‬‫ھﻣﮫ ﺑﺎﺗریھﺎ را ﺑﺎ ﺑﺎﺗریھﺎی ﻧو ﺟﺎﯾﮕزﯾن‬‫ﮐﻧﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ دور ﺑﺎزو ﻗرار دھﯾد‪.‬‬‫ﻣطﻣﺋن ﺷوﯾد ﮐﮫ ﺑدن ﺷﻣﺎ در ﺣﯾن اﻧدازهﮔﯾری‬‫ﮐﺎﻣﻼً ﺑﯽﺣرﮐت و آرام ﺑﺎﺷد‪.‬‬‫راﺣت و ﺑﯽﺣرﮐت ﺑﻧﺷﯾﻧﯾد‪.‬‬‫دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﻗرار دھﯾد ﺑﮫ‬‫ﮔوﻧﮫای ﮐﮫ ﮐف دﺳتھﺎﯾﺗﺎن رو ﺑﮫ ﺑﺎﻻ ﺑوده و‬‫ﺑﺎزوﺑﻧد و ﻗﻠﺑﺗﺎن ھمﺗراز ﺑﺎﺷﻧد‪.‬‬‫اﮔر ﺿرﺑﺎن ﻗﻠب ﺧﯾﻠﯽ ﺿﻌﯾف ﯾﺎ ﻧﺎﻣﻧظﻣﯽ‬‫داﺷﺗﮫ ﺑﺎﺷﯾد‪ ،‬ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫ ﺳﺧﺗﯽ‬‫ﺑﺗواﻧد ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﻌﯾﯾن ﮐﻧد‪.‬‬‫ﺑﮫ ﺑﺧش "ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ‬‫اﻧدازهﮔﯾری ﮐﻧم؟" ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ را ﺧﺎرج ﮐﻧﯾد‪ .‬آﻧﮭﺎ را ﺑﮫ درﺳﺗﯽ‬‫ﻗرار دھﯾد و ﺳﻌﯽ ﮐﻧﯾد ﺑﺎر دﯾﮕر اﻧدازهﮔﯾری‬‫را اﻧﺟﺎم دھﯾد‪.‬‬‫اﮔر ﮐﻠﯾﮫ اﻗداﻣﺎت ذﮐرﺷده را اﻧﺟﺎم دادﯾد اﻣﺎ ﻣﺷﮑﻠﺗﺎن ﺣل ﻧﺷد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬ﺳﻌﯽ ﻧﮑﻧﯾد‬‫اﯾن ﻣﺣﺻول را ﺑﺎز ﯾﺎ ﺗﻌﻣﯾر ﮐﻧﯾد‪ ،‬زﯾرا اﯾن ﮐﺎر ﻣوﺟب ﺑﯽاﻋﺗﺑﺎر ﺷدن ﮔﺎراﻧﺗﯽ آن ﻣﯽﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪20‬‬‫ﻧﮕﮭداری‬‫دﺳﺗﮕﺎه را ﺑﺎز ﻧﮑﻧﯾد‪ .‬اﯾن دﺳﺗﮕﺎه از اﺟزای اﻟﮑﺗرﯾﮑﯽ ظرﯾف و واﺣد ھوای ﭘﯾﭼﯾدهای ﺗﺷﮑﯾل ﺷده اﺳت‬‫ﮐﮫ ﻣﯽﺗواﻧد آﺳﯾب ﺑﺑﯾﻧد‪ .‬اﮔر ﻧﺗواﻧﺳﺗﯾد ﺑﺎ اﺳﺗﻔﺎده از راھﻧﻣﺎی ﻋﯾبﯾﺎﺑﯽ ﻣﺷﮑل دﺳﺗﮕﺎه را ﺣل ﮐﻧﯾد‪ ،‬ﺑﺎ‬‫ﻓروﺷﻧده ﻣﺟﺎز ﺷﮭرﺗﺎن ﯾﺎ ﺑﺧش ﺧدﻣﺎت ﻣﺷﺗرﯾﺎن ﻣﺎ ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬اطﻼﻋﺎت ﻓﻧﯽ‪ ،‬ﻗطﻌﺎت و واﺣدھﺎی‬‫ﯾدﮐﯽ از طرﯾﻖ ﺑﺧش ﺧدﻣﺎت ﻣﺷﺗرﯾﺎن ‪ A&D‬در اﺧﺗﯾﺎر ﻓروﺷﻧدﮔﺎن ﻣﺟﺎز ﻗرار ﻣﯽﮔﯾرﻧد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای ﺧدﻣﺎتدھﯽ طوﻻﻧﯽ ﻣدت طراﺣﯽ و ﺗوﻟﯾد ﺷده اﺳت‪ .‬اﻣﺎ ﺗوﺻﯾﮫ ﻣﯽﺷود ﮐﮫ ھر ‪ ۲‬ﺳﺎل‬‫ﯾﮏﺑﺎر دﺳﺗﮕﺎه را ﺑررﺳﯽ ﮐﻧﯾد ﺗﺎ از ﮐﺎرﮐرد ﺻﺣﯾﺢ و دﻗت آن ﻣطﻣﺋن ﺷوﯾد‪ .‬ﻟطﻔﺎ ً ﺑرای ﺳروﯾس و‬‫ﻧﮕﮭداری دﺳﺗﮕﺎه‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ ﻣﺟﺎز ﯾﺎ ‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫دادهھﺎی ﻓﻧﯽ‬‫ﻧوع‬‫روش اﻧدازهﮔﯾری‬‫ﻣﺣدوده اﻧدازهﮔﯾری‬‫دﻗت اﻧدازهﮔﯾری‬‫ﻣﻧﺑﻊ ﺗﻐذﯾﮫ‬‫ﺗﻌداد اﻧدازهﮔﯾریھﺎ‬‫طﺑﻘﮫﺑﻧدی‬‫آزﻣﺎﯾش ﺑﺎﻟﯾﻧﯽ‬‫‪EMD‬‬‫ﺣﺎﻓظﮫ‬‫ﺷراﯾط ﮐﺎر‬‫ﺷراﯾط ﺣﻣل و ﻧﻘل ‪ /‬ﻧﮕﮭداری‬‫اﺑﻌﺎد‬‫وزن‬‫‪UA-767S‬‬‫اﻧدازهﮔﯾری ﻧوﺳﺎنﺳﻧﺟﯽ‬‫‪ 0‬ﺗﺎ ‪mmHg 299‬‬‫ﻓﺷﺎر‪:‬‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‪ 60 :‬ﺗﺎ ‪mmHg 279‬‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‪ 40 :‬ﺗﺎ ‪mmHg 200‬‬‫‪ 40‬ﺗﺎ ‪ 180‬ﺑﺎر‪ /‬دﻗﯾﻘﮫ‬‫ﺿرﺑﺎن‪:‬‬‫‪mmHg ±3‬‬‫ﻓﺷﺎر‪:‬‬‫‪5%±‬‬‫ﺿرﺑﺎن‪:‬‬‫‪ 4‬ﻋدد ﺑﺎﺗری ‪ 1.5‬وﻟﺗﯽ )‪ LR6 ،R6P‬ﯾﺎ ‪ (AA‬ﯾﺎ‬‫آداﭘﺗور ‪) (TB-233C) AC‬ﻧدارد(‬‫ﺣدود ‪ 700‬ﻣرﺗﺑﮫ ‪) LR6‬ﺑﺎﺗریھﺎی ﻗﻠﯾﺎﯾﯽ(‬‫ﺣدود ‪ 200‬ﻣرﺗﺑﮫ ‪) R6P‬ﺑﺎﺗریھﺎی ﻣﻧﮕﻧزی(‬‫ﺑﺎ ﻣﻘدار ﻓﺷﺎر ‪ ،mmHg 180‬دﻣﺎی اﺗﺎق ‪ 23‬درﺟﮫ ﺳﺎﻧﺗﯽﮔراد‪.‬‬‫ﺗﺟﮭﯾزات ‪ EM‬ﺑﺎ ﺗوان داﺧﻠﯽ )ﺗﺄﻣﯾنﺷده ﺗوﺳط ﺑﺎﺗری( ‪ /‬ﮐﻼس ‪۲‬‬‫)ﺗﺄﻣﯾنﺷده ﺗوﺳط آداﭘﺗور(‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت ﭘﯾوﺳﺗﮫ‬‫ﻣطﺎﺑﻖ اﺳﺗﺎﻧدارد ‪ISO81060-2：2013‬‬‫در ﺑررﺳﯽ اﻋﺗﺑﺎرﺳﻧﺟﯽ ﺑﺎﻟﯾﻧﯽ‪ ،‬از ‪ K5‬ﺟﮭت ﺗﻌﯾﯾن ﻓﺷﺎر ﺧون دﯾﺎﺳﺗوﻟﯽ‬‫ﺑرای ‪ 85‬ﺳوژه اﺳﺗﻔﺎده ﺷد‪.‬‬‫‪IEC 60601-1-2 2014‬‬‫‪ 60‬اﻧدازهﮔﯾری آﺧر‬‫‪ +10‬ﺗﺎ ‪ / °C+40‬رطوﺑت ﻧﺳﺑﯽ ‪ 15‬ﺗﺎ ‪ 85‬درﺻد ‪800 /‬‬‫ﺗﺎ ‪ 1060‬ھﮑﺗوﭘﺎﺳﮑﺎل‬‫‪ -20‬ﺗﺎ ‪ / °C+60‬رطوﺑت ﻧﺳﺑﯽ ‪ 10‬ﺗﺎ ‪ 95‬درﺻد ‪700 /‬‬‫ﺗﺎ ‪ 1060‬ھﮑﺗوﭘﺎﺳﮑﺎل‬‫ﺣدود ‪) 140‬ﻋرض( در ‪) 60‬طول( در ‪) 105‬ﻋﻣﻖ( ﻣﯾﻠﯽﻣﺗر‬‫ﺣدود ‪ 245‬ﮔرم‪ ،‬ﺑدون ﺑﺎﺗریھﺎ‬‫ﻓﺎرﺳﯽ ‪21‬‬‫ﺣﻔﺎظت ورودی‬‫دﺳﺗﮕﺎه‪:‬‬‫‪IP20‬‬‫ﻗطﻌﮫ ﺑﮫﮐﺎررﻓﺗﮫ‬‫ﺑﺎزوﺑﻧد‬‫ﻧوع ‪BF‬‬‫ﻋﻣر ﻣﻔﯾد‬‫‪ 5‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫دﺳﺗﮕﺎه‪:‬‬‫‪ 2‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫ﺑﺎزوﺑﻧد‪:‬‬‫آداﭘﺗور ‪ 5 :AC‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫آداﭘﺗور ‪ AC‬ﯾدﮐﯽ‬‫آداﭘﺗور ﺑرای ﻣﺗﺻل ﮐردن دﺳﺗﮕﺎه ﺑﮫ ﻣﻧﺑﻊ ﺗﻐذﯾﮫ در ﻣﻧزل ﻣﯽﺑﺎﺷد‪ .‬ﻟطﻔﺎ ً ﺑرای ﺧرﯾد آن ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ‬‫‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬آداﭘﺗور ‪ AC‬ﺑﺎﯾد ﺑﮫﺻورت دورهای ﺑررﺳﯽ و ﺗﻌوﯾض ﺷود‪.‬‬‫ﻟطﻔﺎ ً ﺑرای ﺧرﯾد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ ‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫‪TB-233C‬‬‫آداﭘﺗور ‪ AC‬ﺑﺎﯾد ﺑﮫﺻورت دورهای ﺑررﺳﯽ و ﺗﻌوﯾض ﺷود‪.‬‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی آداﭘﺗور ‪ AC‬ﭼﺎپ ﺷده اﺳت‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﺻرﻓﺎ ً ﺑرای اﺳﺗﻔﺎده در ﻣﺣﯾطھﺎی ﺳرﺑﺳﺗﮫ‬‫دﺳﺗﮕﺎه ﮐﻼس دو‬‫ﻓﯾوز ﺣرارﺗﯽ‬‫ﻓﯾوز‬‫ﺑرﭼﺳب دﺳﺗوراﻟﻌﻣل ‪ EC‬دﺳﺗﮕﺎه‬‫ﺑرﭼﺳب ﻣﺟوز ‪ EAC‬دﺳﺗﮕﺎه‬‫ﻗطبھﺎی ﻧری آداﭘﺗور ‪AC‬‬‫ﻟوازم ﺟﺎﻧﺑﯽ ﮐﮫ ﺟدا ﻓروﺧﺗﮫ ﻣﯽﺷوﻧد‬‫اﻧدازه ﺑﺎزو‬‫اﻧدازه ﺑﺎزوﺑﻧد‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن ‪ 31‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 45‬ﺳﺎﻧﺗﯽﻣﺗر‬‫‪CUF-F-LA‬‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 42‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫‪CUF-I‬‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 32‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪CUF-F-A‬‬‫اﻧدازه ﺑﺎزو‪ :‬ﻣﺣﯾط ﺑﺎزو در ﺑﺧش ﻋﺿﻠﮫ دو ﺳر‪.‬‬‫آداﭘﺗور ‪AC‬‬‫ﺗوﺟﮫ‪:‬‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪TB-233C‬‬‫دوﺷﺎﺧﮫ )ﻧوع ﭘرﯾز(‬‫ﻧوع ‪C‬‬‫ﻣﻣﮑن اﺳت ﻣﺷﺧﺻﺎت ﺑدون اطﻼع ﻗﺑﻠﯽ ﺗﻐﯾﯾر داده ﺷوﻧد‪.‬‬‫طﺑﻘﮫﺑﻧدی ‪ IP‬ﻣﯾزان ﺣﻔﺎظت ﻓراھمﺷده ﺗوﺳط ﻣﺣﻔظﮫ ﺗﺟﮭﯾزات ﻣطﺎﺑﻖ ﺑﺎ ‪ IEC 60529‬ﻣﯽﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه در ﺑراﺑر اﺷﯾﺎی ﺧﺎرﺟﯽ ﺟﺎﻣد ﺑﺎ ﻗطر ‪ 12‬ﻣﯾﻠﯽﻣﺗر و ﺑﯾﺷﺗر ﻣﺎﻧﻧد اﻧﮕﺷت ﻣﻘﺎوم ﻣﯽﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه در ﺑراﺑر آب ﻣﻘﺎوم ﻧﯾﺳت‪.‬‬‫ﻓﺎرﺳﯽ ‪22‬‬‫ﯾﺎدداﺷت‬‫ﻓﺎرﺳﯽ ‪23‬‬

A&D Company, Ltd.
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119

OX14 1DY

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409
OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司
中国 上海市浦东新区浦东大道138号永华大厦21楼A室 邮编200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599