ANATEL PAT700

TOTAL ORGANIC CARBON AND CONDUCTIVITY ANALYZER
Fully Compliant with Global Pharmacopeia Requirements : USP, EP, JP

PAT700 TOTAL ORGANIC CARBON & CONDUCTIVITY ANALYZER
LOW COST OF OWNERSHIP
TOC, conductivity and water temperaturefrom just one analyzer•

Can be fully validated for TOC, Conductivity and Temperature
to USP, EP and JP requirements

12 month service interval•

Auto-switching main and standby UV lamps•

UV Detect to ensure UV lamp is working correctly

No chemicals•Designed specifically for pharmaceutical PW and WFImeasurement, this analyzer uses just powerful UV light tooxidize the organics

Fully compliant conductivity meterNo peristaltic pump•

No tubing and pump heads to replace•

PAT700 traps an aliquot for each analysis, so the measurement
is stable and not affected by changes in sample pressure orflow rate

Single sensor to measure TIC and TC•

TC – TIC =TOC•

Stable measurements for >12months

On-line and grab-sample analysis in
one analyzer•

Built-in, 4-bottle grab-sample analyzer

Accelerating Answers

UV Monitoring with auto-switching
Main & Standby UVIMPROVED COMPLIANCE
Fully ICH Q2 compliant•

Not affected by interference from ppm levels of TIC

Complete oxidation
Fully compliant with EP 2.2.44 requirement for complete TOC
oxidation through dynamic end-point detection technology

Increased conductivity
from intermediate species
Dynamic end-point detection.
Final conductivity = TC

Root cause analysis support••Excursion capture feature allows a water sample to be
captured to assist root cause analysis should a TOC excursion
be detected
Built-in grab-sample analyzer for analyzing samples from otherpoints in the water loop

InitialConductivity
= TIC

µS/cm

•µS/cm =TOCTime
UV Lamp off

UV Lamp on

Complete TOC oxidation to EP2.2.44

21 CFR Part 11 Compliant•

Multi-level user access•

Windows credentials (Microsoft Active Directory)•Secure .pdf file export via FTP over Ethernet•

No manual data entry – calibration standards use RFID tags totransfer lot number, expiry date and certified value directly to
the PAT700•

Built-in electronic automated Calibration and System SuitabilitySOPs•

No manual calculations – automated Calibration and System
Suitability Pass/Fail calculationsBuilt-in, automated SOPs

Built-in, automated SOPs

Specifications
TOC

Operating Range

0.5 to 2,000 ppb as Carbon

Display Resolution

0.1 ppb

Accuracy±3 ppb or ±5%, whichever is greater

Repeatability±0.3 ppb or ±1%, whichever is greater

Limit of Detection

0.5 ppb

Maximum Input Conductivity

0.2 μS/cm for all waters, 1.0 μS/cm for all neutral waters,
5.0 μS/cm for water with CO2 as the sole conductive species

Conductivity

Temperature

Physical Specs

Communications

Compliance

New Features

Conductivity

Range 0.05 to 150 μS/cm (@ 25°C)

Display Resolution

0.01 μS/cm

Conductivity Accuracy±2% over full range (uncompensated)

Available Conductivity Reporting Modes

Temperature compensated to 250C, or uncompensated

Available Resisitivity Reporting Mode

Temperature compensated to 250C only

Resistivity

Resistivity Range 0.2 to 18 MΩ-cm (@ 25°C)

Display Resolution

0.01 over full range

Ambient Operating Range

10 to 40°C (50 to 104°F)

Measurement Accuracy±0.5°C

Sample Water Range

1 to 95°C (34 to 203°F)

Display Resolution

0.1 over full range

UV Lamps

2, with UV Detect technology

Interface/Display

Color touchscreen

Maximum Altitude

4,000 m (13,125 ft)

User I/O Wiring

Three, ¾-inch conduit openings or quick disconnect fittings

Standards SystemOnboard, Automated Standards Introduction System (OASIS)

Dimensions

59.7 w X 22.9 d X 25.4 h cm (23.5 X 9 X 10 inches)

Weight13.6 kg (30 lbs)

Sample Inlet Flow Rate Range

60 mL/min to 300 mL/min

Sample Inlet Pressure Range

10 to 100 psi (69 to 690 kPa)

Analog output

3 x 4-20mA outputs, user configurable TOC, Conductivity(uncompensated) and Sample Temperature

Digital output

4 x digital outputs, user configurable (for alarms, etc.)

Digital input

2 x digital inputs (for remote control)

Installation Category

IIPollution Degree

2, IEC 61010-1

CE ComplianceEN 61010-1 and EN 61326

Safety Rating

ETL, conforming to UL 61010-1 and CSA 22.2 No. 61010-1

Enclosure Rating

Conduit version: IP56
Quick connect version: IP46

Release tests,

USP <643>, USP <645>, JP 2.59, EP 2.2.44

CIP

Selectable mode for Clean-In-Place analysis

Dual Stream option

Toggle or programmable stream switching

Excursion sampling

Minimum flow rate to fill excursion bottle = 160 mL/minRouge detection

Identifies oxidation cell contamination from rouging		© 2022 Beckman Coulter, Inc. All rights reserved. Beckman Coulter, the stylized logo, and the Beckman Coulter
product and service marks mentioned herein are trademarks or registered trademarks of Beckman Coulter, Inc. in theUnited States and other countries. All other trademarks are property of their respective owners.
For Beckman Coulter’s worldwide office locations and phone numbers, please visit Contact Us at beckman.com		22.06.5119.PCC