AD-8526 Ethernet Converter

Instruction ManualThe AD-8526 Ethernet Converter can connect the RS-232C interface of a weighing instrument to the
Ethernet (LAN) port of a computer that is not equipped with an RS-232C interface. By using the AD8526, weight data from a weighing instrument can be managed with a computer connected to thenetwork. The weighing instrument can also be controlled by the connected computer.

Unpacking the AD-8526
MAC address

AD8526

Ethernet converter

AC adapter

Positions to affix
AC adapter labels(Example)
NotePlease confirm thatthe AC adapter typeis correct for yourlocal voltage andreceptacle type.

AC adapter labels

Affix them to the AC adapter andthe connector end of the cable.

IP address label (5 labels)

Write a preset IP address andaffix it on the weighing instrument.

IP address protective cover(5 labels)
Affix it on the IP address label.

RS-232C cable(Length: Approx. 2m. 2 pcs)
Note: Only the following cable is included if specified when ordering:
DIN 7-pin to D-Sub 9-pin cable (AX-KO1786-200)
If not specified, the following two communication cables are included:
D-Sub 25-pin to D-Sub 9-pin cable (AX-KO1710-200), D-Sub 9-pin to D-Sub 9-pin cable (AX-KO2466-200)

Specification
1. Specification
Operating environment : -10°C to +40°C

3. RS-232C serial interface
Connector:
D-Sub 9-pin maleTransmission
form:
Asynchronous, bi-directionalthat the adapter type is correct for
AC adapter: Confirm
600, 1200, 2400*, 4800,
the local voltage and power receptacle type
Baud rate: 9600, 19200bps
Power consumption: Approx. 11VA (supplied to the AC adapter) Data format Data bits: 7* or 8 bits
Dimensions:
113(W) x 60(D) x 38(H) mmParity:
Even*, Odd, None
Net weight:
Approximately 250g
Stop bits: 1 bit* or 2 bits
2. Ethernet interface
Connector:
RJ45* : Factory settings
Protocol:TCP/IP

Weighing Instrument Cables

Required AD-8526 Cable by Weighing Instrument
Communications cable (Length 2 m)
Weighing instrument
Interface option
GX, GF, GX-K, GF-K, GP, GR, HR,
None (D-Sub 25-pin, standard accessory)
AX-KO1710-200
MCNone (D-Sub 9-pin, standard accessory)
EK-i, EW- i, FC- i, FC-S i, GH, HR- i
AX-KO2466-200
None (D-Sub 9-pin, standard accessory)
AD-4212C
AD4212C standard accessory
None (DIN 7-pin, standard accessory)
HV-G, HV-WP, HW-G, HW-WP
AX-KO1786-200
OP-03(DIN7-pin)
FG
AX-KO1786-200
OP-23 (DIN 8-pin)
FG-L, FG-M
AX-KO1786-200
Please refer to our website for the latest information.
1WMPD4001294D

Connecting the Weighing Instrument to a Computer

CautionContact your network manager before connecting the AD-8526 to a network. The AD-8526 may
generate a network error. A&D assumes no responsibility for any errors that may be caused.
Affix the AC adapter labels to the AC adapter and the connector end of the cable.
Set the IP address and subnet mask to the AD-8526 one at a time.
The factory default IP address is 172.16.100.2. Do not duplicate an IP address.
The IP address cannot be returned to the factory setting.
AD-8526
We recommend that you write the IP address on the supplied label.

Preparation

Step 2
Please download each software from A&D website（https://www.aandd.jp/products/software/software.html）for below.
(1) Setting procedure “WinCT-Plus” instruction manual(2) IP address setting software “Device Installer”(3) Data acquisition software “WinCT-Plus”

Step 1Connect the AC adapter to the AD-8526.

Step 2Connect the AD-8526 and the computer with a cross
cable directly or connect them with a hub and straightcables.

AC adapter
Step1
Cross cable

Straight cable

Step 3 Input the IP addresses and subnet masks of the AD-8526
and computer. For the setting procedure, refer to theAD-8526“WinCT-Plus” instruction manual.
Communication error may occur if set incorrectly.
Step 6
Step 4 Write the IP address on the IP address label and affix iton the AD-8526 (and the weighing instrument ifnecessary) where you can see the address easily.
Step 5 Install the RsMulti data acquisition software to download the“WinCT-Plus” instruction manual.PDF form A&D website.
（https://www.aandd.jp/products/software/software.html）
Step 6 Connect the accessory RS-232C cable between the RS-232C
interface of the weighing instrument and the AD-8526’s port.

Step 2Computer
Weighinginstrument

Step 6
RS-232C cable

Data Communication Software

The RSMulti data acquisition software can be launched on a Windows PC. Menu: [Start][Program][A&D WinCT-Plus][RsMulti]. Refer to the Manual PDF file in the A&D WinCT-Plus for details on theoperation.
This software can acquire data from multiple devices connected by LAN or RS-232C.
The software can control these devices with commands.
The software can acquire data transmitted from devices. Example: When the PRINT key is pressed on thebalance, data is output and is acquired with the computer.
Stored data can be used with Excel. (Install Microsoft Excel before use.)

ExampleTwo balances are connected.
IP address
Subnet mask
Computer 172.16.100.1
GX-600172.16.100.2 255.255.0.0
GP-30K
172.16.100.3
HubComputer
172.16.100.1

172.16.100.2
GX-600172.16.100.3
GP-30KA sample window of RsMulti