IOL PORTFOLIO

Hydrophobic & Hydrophilic Solutions

EVERY PATIENT ISUNIQUEOUR SOLUTIONS TOOService and support throughout your cataract surgery pathway.
BVI has grown to be a highly regarded ophthalmic device manufacturer offering abroad portfolio of products, including monofocal and premium Intraocular Lenses(IOLs), a full range of ophthalmic single use consumables, surgical fluids, phaco
systems and custom procedure packs. BVI provides innovative and high quality
products that perform consistently and predictably for surgeons across the globe.

For over three decades we have been leading the way in the design anddevelopment of IOLs, and we continue to prioritize three key areas:· Striving to offer high-performance optical solutions.
· Meeting the strictest requirements for medical device directives and regulations.
· Focusing to improve the quality of sight and therefore, the quality of life.

Unburdened by legacy or bureaucracy, we have developed our strategy around asimple concept — taking pride in delivering innovative solutions for our physicians
and patients, based on their needs.

We trust and empower our associates to make decisions and solve problems
because collaboration drives us. Valuing agility, simplicity, and transparency, we staycommitted to listening to our customers, delivering for our patients, and keeping
the future in focus.

3

Product Families

Product Families

GFY HYDROPHOBIC MATERIALHYDROPHILIC MATERIALPREMIUM TRIFOCAL - FINE TECHNOLOGYPREMIUM TRIFOCAL - FINE TECHNOLOGY

FINE VISION HP

FINE VISION HP

TRIFOCAL OPTICTRIFOCAL OPTIC

FINE VISIONTRIFOCAL OPTIC

PREMIUM MONOFOCAL - ISOFOCAL TECHNOLOGYISOPURE

FINE VISIONFINE VISIONTRIFOCAL OPTICTRIFOCAL OPTICMONOFOCAL TORICANKORISISOPUREMONOFOCAL OPTICMONOFOCAL TORIC

PODE YEMONOFOCAL OPTICMONOFOCALMICROPURE

MICROPURE

PODE YEMONOFOCAL OPTICMONOFOCAL OPTICMONOFOCAL OPTICINJECTION SYSTEM

Premium

Medicel Accuject / Viscoject

MICRO Platform

POD Platform

MICRO Platform

POD Platform

Note :The intraocular lenses are not FDA approved. Please check the lens availability with your sales representative.
The ISOPURE 123 and MICROPURE 123 lenses are delivered preloaded in a cartridge, which is simply clipped to the Single-Use Injector 1.2.3.

Note :

Premium. If you need Single-Use Injector 1.2.3. Premium, please check the availability of the products on your market with your sales representative.

The intraocular lenses are not FDA approved. Please check the lens availability with your sales representative.

5

Injection Systems

Diopter Range Overview

1

Optic

Material-10D-1D 0D

6D

10D

TRIFOCAL TORICHYDROPHOBIC

FINEVISION HP TORIC (POD FT 49P)2

TRIFOCALHYDROPHOBIC

FINEVISION HP (POD F GF)

TRIFOCAL TORICHYDROPHILICFINEVISION TORIC (POD FT)2

TRIFOCAL

Optic

Power range

30D 31D

35D
TRIFOCAL TORICHYDROPHOBIC

FINE VISION HP

TRIFOCALHYDROPHOBIC

FINE VISION HP

TRIFOCAL TORICHYDROPHILIC

FINE VISIONTRIFOCALHYDROPHILICFINEVISION (POD F)TRIFOCAL OPTICTRIFOCAL OPTICTRIFOCAL OPTIC

FINE VISIONTRIFOCAL OPTIC

PREMIUM MONOFOCALHYDROPHOBICMONOFOCAL TORICHYDROPHOBICMONOFOCAL TORICISOPURE

FINE VISIONTRIFOCAL OPTICISOPURE 123
PODEYE TORIC (POD T 49P)2
ANKORIS3

HYDROPHILIC

Model

Injection system
Viscoject
Bio 1.8 Accuject Accuject Accuject
Accuject
2.0
2.1
2.2
1.8

FINEVISION (MICRO F)

HYDROPHILIC

Brand

Material

PREMIUMMONOFOCALHYDROPHOBICMONOFOCAL TORICHYDROPHOBICMONOFOCAL TORICHYDROPHILICPOD FT 49P

≤ 35D

≤ 35D

POD F GF

≤ 24.5D

≤ 35D

≤ 35D

POD FT≤ 24.5D

≤ 35D

≤ 35D

≤ 35D

≤ 35D

≤ 35D

≤ 24.5D

≤ 35D

≤ 35D

≤ 35D

≤ 35D

≤ 35D

≤ 30D

≤ 30D

MICRO F≤ 24.5D

POD F

ISOPUREISOPUREISOPUREISOPURE 123

PODE YE

POD T 49P

≤ 24.5D1

123
Premium

PODEYEMONOFOCALHYDROPHOBIC

MICROPURE
MICROPURE 123

MONOFOCAL OPTICANKORISMONOFOCAL OPTIC

PODE YEMONOFOCAL OPTICMONOFOCALHYDROPHOBIC

MICROPUREMONOFOCAL OPTICMICROPUREMONOFOCAL OPTIC

1

Refer to our website for updates

2

Cylinder power: 1.00 - 1.50 - 2.25 - 3.00 - 3.75 - 4.50 - 5.25 - 6.00D

3

Cylinder power : 1.50 - 2.25 - 3.00 - 3.75 - 4.50 - 5.25 - 6.00D

Please check the lens availability with your sales representative

ANKORIS≤ 24.5D

≤ 30D

≤ 30D

PODEYE≤ 24.5D

≤ 35D

≤ 35D

≤ 35D

≤ 35D

≤ 35D

MICROPURE≤ 24.5D1

MICROPURE 123

Only Accuject 1.8
Please check the lens availability with your sales representative
1

7

TRIFOCAL TORIC

TrifocalToric
9

TRIFOCAL TORICTORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

FINE VISION HP
TRIFOCAL OPTIC

11.40mmOVERALL
DIAMETERModel

POD FT 49P

Material

GFY Hydrophobic Acrylic1

Overall diameter

CONVOLUTED &APODIZED DESIGNASPHERIC SURFACE

11.40mm

Optic diameter

6.00mmOptic

Biconvex Aspheric Toric Trifocal

Haptic design

POD (Double C-loop) with Ridgetech® & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.53

Abbe number

42

Additional power (IOL plane)+1.75D & +3.50D

Injection system
ANTERIORDIFFRACTIVE
SURFACE

Medicel Accuject 2.1/2.2

Spherical power4+10D to +35D (0.5D steps)

Cylinder power (IOL plane)

1.00 - 1.50 - 2.25 - 3.00 - 3.75 - 4.50 - 5.25 - 6.00D

4Interferometry

Suggested A constant2
Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A

119.40

Haigis3: a0; a1; a2

Trifocal
Toric
Hydrophobic

Cylinder power at IOL plane
Cylinder power at corneal plane

511.70; 0.4; 0.1

POD FT
49P 1.0

POD FT
49P 1.5

POD FT49P 2.25

POD FT49P 3.0

POD FT49P 3.75

POD FT49P 4.5

POD FT49P 5.25

POD FT49P 6.0

1.00D

1.50D

2.25D

3.00D

3.75D

4.50D

5.25D

6.00D

0.68D

1.03D

1.55D

2.06D

2.57D

3.08D

3.60D

4.11D

The BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical and cylinder powers with your sales representative. | 5 Savini G., J Cataract Refract Surg 2013; 39:1900 –1903.

TORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

11

TRIFOCAL TORICTORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

FINE VISIONTRIFOCAL OPTIC

11.40mmOVERALL
DIAMETERCONVOLUTED &APODIZED DESIGNASPHERIC TORICSURFACEModel

POD FT

Material

26% Hydrophilic Acrylic

Overall diameter

11.40mm

Optic diameter

6.00mmOptic

Biconvex Aspheric Toric Trifocal

Haptic design

POD (Double-C-loop) haptic design & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.46

Abbe number

58

Additional power (IOL plane)
ANTERIORDIFFRACTIVE
SURFACE+1.75D & +3.50D

Injection system

Medicel Accuject 2.0 up to 24.5D and Medicel Accuject 2.1/2.2 up to 35D

Spherical power+6D to +35D (0.5D steps)

3

Cylinder power (IOL plane)

1.00 - 1.50 - 2.25 - 3.00 - 3.75 - 4.50 - 5.25 - 6.00D

3Interferometry

Suggested A constant1

Trifocal
Toric
Hydrophilic

Hoffer Q: pACD

5.59

Holladay 1: Sf

1.83

Barrett: LF

1.86

SRK/T: A

118.95

Haigis : a0; a1; a2

1.36; 0.4; 0.1

2

1

POD FT
1.0

POD FT
1.5

POD FT
2.25

POD FT
3.0

POD FT
3.75

POD FT
4.5

POD FT
5.25

POD FT
6.0

Cylinder power at IOL plane

1.00D

1.50D

2.25D

3.00D

3.75D

4.50D

5.25D

6.00D

Cylinder power at corneal plane4

0.68D

1.03D

1.55D

2.06D

2.57D

3.08D

3.60D

4.11D

V
 alues estimated only: surgeons are recommended to personalize their A-constant based on their surgical techniques and equipment, experience withthe lens model and postoperative results. | 2 Not optimized. | 3 Please check the availability of spherical and cylinder powers with your sales representative.
| 4 Savini G., J Cataract Refract Surg 2013; 39:1900 –1903.

TORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

13

TRIFOCAL

Trifocal

15

FINE VISION HP

11.40mmOVERALL
DIAMETERCONVOLUTED &APODIZED DESIGNASPHERIC SURFACEModel

POD F GF

Material

GFY Hydrophobic Acrylic1

Overall diameter

11.40mm

Optic diameter

6.00mmOptic

Biconvex Aspheric Trifocal

Haptic design

POD (Double C-loop) with Ridgetech® & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.53

Abbe number

42

Additional power (IOL plane)
ANTERIORDIFFRACTIVE
SURFACE+1.75D & +3.50D
Medicel Accuject 2.0 up to 24.5D
Medicel Accuject 2.1/2.2 up to 35D

Injection system
Spherical power4+10D to +35D (0.5D steps)
Interferometry

Suggested A constant2

Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A

119.40

Haigis : a0; a1; a2
3

TrifocalHydrophobic

1

TRIFOCALTRIFOCAL OPTIC

1.70; 0.4; 0.1

The BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical powers with your sales representative.

17

FINE VISION

10.75mmOVERALL
DIAMETERCONVOLUTED &APODIZED DESIGNASPHERIC SURFACE

Model

MICRO F

Material

25% Hydrophilic Acrylic

Overall diameter

10.75mm

Optic diameter

6.15mm

Optic
Haptic design

Biconvex Aspheric Trifocal
MICRO (closed loop quadripode) & Posterior Angulated Haptic

Filtration
6.15mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.46

Abbe number

58

Additional power (IOL plane)

ANTERIORDIFFRACTIVE
SURFACE+1.75D & +3.50D

Injection system

Medicel Viscoject Bio 1.8 / Accuject 1.8 up to 24.5D
Medicel Accuject 2.0/2.1/2.2 up to 35D

Spherical power3+10D to +35D (0.5D steps)
Interferometry

Suggested A constant1
Hoffer Q: pACD

5.35

Holladay 1: Sf

1.60

Barrett: LF

1.78

SRK/T: A

118.80

Haigis : a0; a1; a2
2

TrifocalHydrophilic

1

TRIFOCALTRIFOCAL OPTIC

1.36; 0.4; 0.1

Values estimated only: surgeons are recommended to personalize their A-constant based on their surgical techniques and equipment, experience withthe lens model and postoperative results. | 2 Not optimized. | 3 Please check the availability of spherical powers with your sales representative.

19

FINE VISION

11.40mmOVERALL
DIAMETERCONVOLUTED &APODIZED DESIGNASPHERIC SURFACE

Model

POD F

Material

26% Hydrophilic Acrylic

Overall diameter

11.40mm

Optic diameter

6.00mm

Optic
Haptic design

Biconvex Aspheric Trifocal
POD (Double-C-loop) haptic design & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.46

Abbe number

58

Additional power (IOL plane)
ANTERIORDIFFRACTIVE
SURFACE+1.75D & +3.50D

Injection system

Medicel Accuject 2.0 up to 24.5D
Medicel Accuject 2.1/2.2 up to 35D

Spherical power3+6D to +35D (0.5D steps)
Interferometry

Suggested A constant1
Hoffer Q: pACD

5.59

Holladay 1: Sf

1.83

Barrett: LF

1.86

SRK/T: A

118.95

Haigis : a0; a1; a2
2

TrifocalHydrophilic

1

TRIFOCALTRIFOCAL OPTIC

1.36; 0.4; 0.1

Values estimated only: surgeons are recommended to personalize their A-constant based on their surgical techniques and equipment, experience withthe lens model and postoperative results. | 2 Not optimized. | 3 Please check the availability of spherical powers with your sales representative.

21

PREMIUM MONOFOCAL

Premium
Monofocal
23

POLYNOMIALSURFACE DESIGN

Model

ISOPURE 123

Material

GFY Hydrophobic Acrylic1

Overall diameter

10D to 24.5D: 11.00mm
25D to 30D: 10.75mm

Optic diameter

10D to 24.5D: 6.00mm
25D to 30D: 5.75mm
Polynomial Surface Design

Optic
Haptic design

MICRO (closed loop quadripode) & Posterior Angulated Haptic

Filtration

UV & Blue Light

Refractive index

1.53

Abbe number

42

Injection system

SINGLE-USE INJECTOR 1.2.3 PREMIUM

Spherical power4+10D to +30D (0.5D steps) Cartridge with PRS® technology
Interferometry

Suggested A constant2

Preloaded
Premium
Monofocal
Hydrophobic

PREMIUM MONOFOCALISOPURE

Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A

119.40

Haigis : a0; a1; a2

1.70; 0.4; 0.1

3

ISOPURE1

Overall diameter

10D to 24.5D: 11.00mm - 25D to 35D: 10.75mm

Optic diameter

10D to 24.5D: 6.00mm - 25D to 35D: 5.75mm

Injection system

Medicel Accuject 1.8 up to 24.5D - Medicel Accuject 2.0/2.1/2.2 up to 35D

Spherical power4+10D to +30D (0.5D steps) - +31D to +35D (1D steps)

The BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical powers with your sales representative.

25

MONOFOCAL TORIC

Monofocal
Toric
27

TORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

PODE YEISOPUREMONOFOCAL OPTIC

BICONVEX ASPHERICTORIC MONOFOCALSURFACE

Model

PODEYE TORIC

Material

GFY Hydrophobic Acrylic1

Overall diameter

11.40mm

Optic diameter

6.00mmOptic

Biconvex Aspheric Toric Monofocal

Haptic design

POD (Double-C-loop) haptic design & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.53

Abbe number

42

Injection system

Medicel Accuject 2.1 / 2.2

Spherical power4+6D to +30D (0.5D steps)

Cylinder power (IOL plane)4

1.00 - 1.50 - 2.25 - 3.00 - 3.75 - 4.50 - 5.25 - 6.00DInterferometry

Suggested A constant2
Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A

MONOFOCAL TORIC

11.40mmOVERALL
DIAMETER

119.40

Haigis3: a0; a1; a2

1.70; 0.4; 0.1

PODEYE PODEYE PODEYE PODEYE PODEYE PODEYE PODEYE PODEYETORIC 1.0 TORIC 1.5 TORIC 2.25 TORIC 3.0 TORIC 3.75 TORIC 4.5 TORIC 5.25 TORIC 6.0

Monofocal
Toric
Hydrophobic

Cylinder power at IOL plane
Cylinder power at corneal plane

5

1

1.00D

1.50D

2.25D

3.00D

3.75D

4.50D

5.25D

6.00D

0.68D

1.03D

1.55D

2.06D

2.57D

3.08D

3.60D

4.11D

The BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical and cylinder powers with your sales representative. | 5 Savini G., J Cataract Refract Surg 2013; 39:1900 –1903.

TORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

29

TORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

ANKORISMONOFOCAL OPTIC

BICONVEX ASPHERICTORIC MONOFOCALSURFACE

Model

ANKORIS

Material

26% Hydrophilic Acrylic

Overall diameter

11.40mm

Optic diameter

6.00mmOptic

Biconvex Aspheric Toric Monofocal

Haptic design

POD (Double-C-loop) haptic design & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.46

Abbe number

58

Injection system

Medicel Accuject 2.0 up to 24.5D & Medicel Accuject 2.1/2.2 up to 30D

Spherical power+6D to +30D (0.5D steps)

3

Cylinder power (IOL plane)

1.50 - 2.25 - 3.00 - 3.75 - 4.50 - 5.25 - 6.00D

3Interferometry

Suggested A constant1
Hoffer Q: pACD

5.59

Holladay 1: Sf

1.83

Barrett: LF

1.86

SRK/T: A

118.95

Haigis : a0; a1; a2

1.36; 0.4; 0.1

2

Monofocal
Toric
Hydrophilic

Cylinder power at IOL plane
Cylinder power at corneal plane

4

1

MONOFOCAL TORIC

11.40mmOVERALL
DIAMETERANKORIS
1.5

ANKORIS
2.25

ANKORIS
3.0

ANKORIS
3.75

ANKORIS
4.5

ANKORIS
5.25

ANKORIS
6.0

1.50D

2.25D

3.00D

3.75D

4.50D

5.25D

6.00D

1.03D

1.55D

2.06D

2.57D

3.08D

3.60D

4.11D

Values estimated only: surgeons are recommended to personalize their A-constant based on their surgical techniques and equipment, experience withthe lens model and postoperative results. | 2 Not optimized. | 3 Please check the availability of spherical and cylinder powers with your sales representative.
| 4 Savini G., J Cataract Refract Surg 2013; 39:1900 –1903.

TORIC CALCULATOR:
WWW.PHYSIOLTORIC.EU - SEE P. 40

31

MONOFOCAL

Monofocal

33

MICROPURE
MONOFOCAL OPTIC

Model

MICROPURE 123

Material

GFY Hydrophobic Acrylic1
0D to 24.5D: 11.00mm
25D to 30D: 10.75mm
0D to 24.5D: 6.00mm
25D to 30D: 5.75mm

Overall diameter
Optic diameter
Optic

Biconvex Aspheric Monofocal

Haptic design

MICRO (closed loop quadripode) & Posterior Angulated Haptic

Filtration

UV & Blue Light

Refractive index

1.53

Abbe number

42

Injection system

SINGLE-USE INJECTOR 1.2.3 PREMIUM

Spherical power4

0D to +9D (1D steps) & +10D to +30D (0.5D steps)
Cartridge with PRS technology
Interferometry

Suggested A constant2
Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A

119.40

Haigis : a0; a1; a2
3

Preloaded
Monofocal
Hydrophobic

11.70; 0.4; 0.1

MONOFOCALBICONVEX ASPHERICMONOFOCAL
SURFACEThe BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical powers with your sales representative.

35

MICROPURE
MONOFOCAL OPTIC

Model

MICROPURE

Material

GFY Hydrophobic Acrylic1-10D to 24.5D: 11.00mm
25D to 35D: 10.75mm-10D to 24.5D: 6.00mm
25D to 35D: 5.75mm

Overall diameter
Optic diameter
Optic

Biconvex Aspheric Monofocal

Haptic design

MICRO (closed loop quadripode) & Posterior Angulated Haptic

Filtration

UV & Blue Light

Refractive index

1.53

Abbe number

42

Injection system

Medicel Accuject 1.8 up to 24.5D
Medicel Accuject 2.0/2.1/2.2 up to 35D

Spherical power4-10D to +9D (1D steps)+10D to +30D (0.5D steps)+31D to +35D (1D steps)
Interferometry

Suggested A constant2
Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A
Haigis3: a0; a1; a2

Monofocal
Hydrophobic

1

119.401.70; 0.4; 0.1

MONOFOCALBICONVEX ASPHERICMONOFOCAL
SURFACEThe BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical powers with your sales representative.

37

PODE YEMONOFOCAL OPTIC

BICONVEX ASPHERICMONOFOCAL
SURFACE

Model

PODEYE

Material

GFY Hydrophobic Acrylic1

Overall diameter

11.40mm

Optic diameter

6.00mmOptic

Biconvex Aspheric Monofocal

Haptic design

POD (Double-C-loop) haptic design & Posterior Angulated Haptic

Filtration
6.00mmOPTIC
DIAMETER

UV & Blue Light

Refractive index

1.53

Abbe number

42

Injection system

Medicel Accuject 2.0 up to 24.5D
Medicel Accuject 2.1/2.2 up to 35D

Spherical power4+10D to +30D (0.5D steps)
0D to +9D & +31D to +35D (1D steps)
Interferometry

Suggested A constant2
Hoffer Q: pACD

5.85

Holladay 1: Sf

2.06

Barrett: LF

2.09

SRK/T: A

119.40

Haigis3: a0; a1; a2

Monofocal
Hydrophobic

11.70; 0.4; 0.1

MONOFOCAL

11.40mmOVERALL
DIAMETER

The BVI GFY® is patented since 2010. Patent number: EP1830898. | 2 Values estimated only: surgeons are recommended to personalize their A-constantbased on their surgical techniques and equipment, experience with the lens model and postoperative results. | 3 Not optimized. | 4 Please check the availability of spherical powers with your sales representative.

39

www.physioltoric.eu

Online Toric Calculator
with Abulafia-Koch regression formula
How to achieve the most accurate correction for your astigmatic patients?
Our goal is to assist surgeons with precise and reliable IOL calculations. The new calculation method informs physicians about
the appropriate toric IOL model and as such improves toric outcomes in astigmastic patients.

What are the new features?
1

Abulafia-Koch regression Formula, which reportedly theoretically accounts for posterior corneal astigmatism. This calculation
method uses the standard keratometry measurements (anterior K values) and estimates the total corneal astigmatism basedon the Abulafia-Koch regression Formula to improve the prediction of postoperative astigmatic outcome. Calculation using
the Standard K method is still possible.

2

HELP-button at each bloc that will help you understand and fill in each parameter.

3

Predictive patient-specific effective lens position (ELP)The calculator still offers the possibility to use the Standard K calculation method as with the previous version.

Toric Calculator

www.physioltoric.eu

Abulafia-Koch
regression Formula

1

HELP-button

2

Predictive patientspecific effective
lens position

341

This page is intentionally left blank

This page is intentionally left blank

43

IOL PORTFOLIO
Intraocular Solutions Overview
Contact Information:www.bvimedical.com/customer-support/

BVI and all other trademarks (unless noted otherwise) are property of BVI. BVI ©2023

1586019-02