LCR Data Logger
User Manual

Notices
The information contained in this document is provided “as is,” and is subject to change, without notice, in
future editions.
This document contains proprietary information that is protected by copyright. All rights are reserved. No partof this document may be photocopied, reproduced, or translated to another language without the prioragreement and written consent from LCR Research Ltd.
© LCR Research Ltd. 2016

Warranty
LCR Research disclaims all warranties, either express or implied, with regard to this manual and any information
contained herein, including but not limited to the implied warranties of merchantability and fitness for aparticular purpose. LCR Research shall not be liable for errors or for incidental or consequential damages inconnection with the furnishing, use, or performance of this document or of any information contained herein.

Exclusive Remedies
The remedies provided herein are the Buyer’s sole and exclusive remedies. LCR Research shall not be liable forany direct, indirect, special, incidental, or consequential damages, whether based on contract, tort, or any otherlegal theory. Such excluded damages shall include, but are not limited to: costs of removal and installation,
losses sustained as the result of injury to any person, or damage to property.

LCR Data Logger User Manual - ver 1.04

Contents
Introduction...........................................................................................................................................................................................1
System Requirements ................................................................................................................................................................1

Installing Software .........................................................................................................................................................................2
Connecting to Device...................................................................................................................................................................8
Logging Measurement Results ....................................................................................................................................... 10
Setting Up Measurement Parameters ..................................................................................................................... 14
Measurement Setup................................................................................................................................................................ 14
Sorting Setup ............................................................................................................................................................................. 17

Setting Up System Parameters ....................................................................................................................................... 19
System Setup ............................................................................................................................................................................. 19
Device Information .................................................................................................................................................................. 20
Self-Calibration ......................................................................................................................................................................... 20
GUI ................................................................................................................................................................................................. 21

Using Multiple Component Identification Function ................................................................................ 22
Parameters Setup ..................................................................................................................................................................... 22
Visual Multi-Component Measurement .......................................................................................................................... 25
Non-Visual Multi-Component Measurement ............................................................................................................... 35

Uninstalling Software ............................................................................................................................................................... 41

LCR Data Logger User Manual - ver 1.04

Introduction
The LCR Data Logger is an application software that provides graphical user interface (GUI) for device
programming and data logging. It is available for free on the LCR Research website(http://www.lcrresearch.com/downloads).

System Requirements
Before installing the LCR Data Logger software, ensure that your PC meets the following minimum system
requirements:
32-bit OS Platforms
MemoryHard Disk Space
Supported Models
Required Accessory
Prerequisite

1

Windows 7, 8 or 10
512 MB or higher
1GB free disk space
LCR Pro1 portable LCR meter ("LCR Pro1")
LCR Link1 communication dongle ("LCR Link1")
Microsoft Excel 2007 or higher

LCR Data Logger User Manual - ver 1.04

Installing Software
Please take the following steps to install the LCR Data Logger software on PC:
1.
2.
3.
4.

Download the LCR Data Logger software from http://www.lcrresearch.com/downloads.
Unzip the file LCR Data Logger 0.31.zip.
Open the folder LCR Data Logger 0.31.
Run setup.bat as Figure 2-1 shows.

Figure 2-1: Run setup.bat
5. It requests the permission to install FTDI CDM driver as Figure 2-2 shows. Click the Yes button tocontinue.

Figure 2-2: Request to Install FTDI CDM Driver

2

LCR Data Logger User Manual - ver 1.04

6. Click the Extract button to install FTDI CDM driver as Figure 2-3 shows.

Figure 2-3: Extract FTDI CDM Driver
7. Click the Next button as Figure 2-4 shows.

Figure 2-4: Install FTDI CDM Driver

3

LCR Data Logger User Manual - ver 1.04

8. Select the I accept this agreement option, then click the Next button as Figure 2-5 shows if you agree
the FTDI license agreement.

Figure 2-5: FTDI License Agreement
9. Click the Finish button as Figure 2-6 shows after the FTDI CDM driver is installed.

Figure 2-6: Finish FTDI CDM driver installation

4

LCR Data Logger User Manual - ver 1.04

10. Click the Next button as Figure 2-7 shows to start LCR Data Logger Setup Wizard.

Figure 2-7: LCR Data Logger Setup Wizard
11. Select the I Agree option, then click the Next button as Figure 2-8 shows if you agree the LCR Research
End-User License Agreement.

Figure 2-8: LCR Research End-User License Agreement

5

LCR Data Logger User Manual - ver 1.04

12. Select the installation folder and click the Next button. The default folder is C:\Program Files (x86)\LCR
Research\LCR Data Logger\.

Figure 2-9: Select installation Folder
13. Click the Next button as Figure 2-10 shows.

Figure 2-10: Start LCR Data Logger Installation

6

LCR Data Logger User Manual - ver 1.04

14. It requests the permission to install LCR Data Logger software as Figure 2-10 shows. Click the Yes button
to continue.

Figure 2-10: Request the Permission to Install LCR Data Logger Software
15. Click the Close button as Figure 2-11 shows when LCR Data Logger software is installed.

Figure 2-11: Finish LCR Data Logger Software installation

NOTE
If an old version of LCR Data Logger software has already been installed in your PC, you need to uninstall it
before installing a newer version. Please refer to the "Uninstalling Software" section on page 38 for detailed
un-installation procedures.

7

LCR Data Logger User Manual - ver 1.04

Connecting to Device
Please take the following steps to connect the LCR Pro1 to the LCR Data Logger software:
1. Plug in the LCR Link1 to a USB port on your PC.

NOTE
The LCR Link1 is an optional accessory of LCR Pro1. It is a dongle style communication module that links the LCR
Pro1 to PC. Using the LCR Link1, you can log data and charge the device simultaneously.
2. Use a USB cable to connect the LCR Pro1 to the LCR Link1 as Figure 3-1 shows.

Figure 3-1: Connect LCR Pro1 to PC
3. Run the LCR Data Logger software by opening LCR_PC_IF.exe from the location where the application is
installed. The default location is at Windows (C:) > Program Files (x86) > LCR Research > LCR Data
Logger > LCR_PC_IF.exe. Alternatively, you can open LCR_PC_IF.exe by double clicking the shortcut icon
of LCR Data Logger on your PC desktop as Figure 3-2 shows.

Figure 3-2: LCR Data Logger Shortcut Icon

8

LCR Data Logger User Manual - ver 1.04

4. The LCR Data Logger software will connect to the LCR Pro1 automatically. Once connected, you will see aduplicated device display from the bottom right on GUI as Figure 3-3 shows.

Figure 3-3: Message that Device is Connected
5. If the software hasn't connected to the device, GUI shows the "DEVICE DISCONNECTED, WAITING FOR
CONNECTION" message as Figure 3-4 shows.

Figure 3-4: Message that Device not Connected

NOTE
If the "DEVICE DISCONNECTED, WAITING FOR CONNECTION" message stays for more than 10 seconds since the
LCR Pro1 has connected to PC, please try to close the software then reopen it to establish the connection.
9

LCR Data Logger User Manual - ver 1.04

Logging Measurement Results
The logging settings can be configured from the Logging tab on the LCR Data Logger software. Figure 4-1
shows a typical screen in the logging tab.

Figure 4-1: Logging Tab

NOTE
In each tab, you will see some buttons or dropdown menus are grayed out. It means they are not available forthe current situation. For example, the Export button is grayed out during logging period. After you stop logging
the data, it becomes active for you to save the results.
The Logging Autoset button can reset the logging settings to default which are:
Logging Mode: Continuous
Logging Interval (second): 1
Data Filter: Off
The Logging Mode includes three options:•••Continuous - Continuously log the measurement results
Up to X Count - Log the measurement results X times
Press R Key - Log the measurement result every time when the R key on PC is pressed

The Logging Interval (second) sets how fast the measurement results will be recorded. It can be selected from
1 second to 100 seconds with 1 second per step.

10

LCR Data Logger User Manual - ver 1.04

When the Logging Mode is selected as Up to X Count, you can go to Logging Count to select how manytimes the results need to be recorded. The number can be selected from 1 to 65535.
The Data Filter is used to set a threshold to prevent undesired data from being logged due to bad contact,switching components, etc. It has 11 settings which are OFF, 1%, 2%,..., 9%, 10%. If OFF is selected, it records
every result getting from the device. If a percentage is selected, it only records the results which are within thedefined percentage range comparing to the previous 2 results.
For example, if 3% is selected on Data Filter and the device gets a series of results as 1uF, 1.05uF, 0.95uF, 0.99uF,
1.01uF, 1uF, 0.99uF, 1.01uF, 1uF, 1.05uF, 1uF. The software will only record the stable results which are1uF,
0.99uF, 1.01uF, 1uF. The results that are crossed out below vary more than 3% comparing to their previous 2
results, therefore, they are not recorded.
1uF, 1.05uF, 0.95uF, 0.99uF, 1.01uF, 1uF, 0.99uF, 1.01uF, 1uF, 1.05uF, 1uF
The Component Designator box is used to add a reference for the component under testing. The reference
name and/or number will be recorded in the table with other testing parameters and results during data logging
period.
When the Start Logging button is clicked, the LCR Data Logger software starts to log data. Figure 4-2 shows atypical logging screen when measuring a 1uF capacitor. As you can see, the testing parameters and results arelogged in the spreadsheet. The real time measurement result is displayed on the bottom right of the screen.

Figure 4-2: Logging Screen in Measurement
During the logging period, you can pause the logging by clicking the Hold button. Once it is clicked, the button
message has been changed to Resume. You can resume the logging by clicking the Resume button.

11

LCR Data Logger User Manual - ver 1.04

To stop logging, please click the Stop Logging button. Once it is clicked, you will see the Clear button andExport button become active. Click the Clear button to clear all the logging data. Click the Export button tosave the logging data as a spreadsheet to a directory you selected.
Once the Export button is clicked, a dialog box prompts to ask for a location for saving the test results as Figure
4-3 shows. Press the Save button on the dialog box to save the file. The default location is at Windows(C:) >LCR_DATA.

Figure 4-3: Save Logging Results
After the test results are saved, a dialog box prompts to indicate the file is saved successfully as Figure 4-4
shows.

12

LCR Data Logger User Manual - ver 1.04

Figure 4-4: Successfully Saved Logging Results

13

LCR Data Logger User Manual - ver 1.04

Setting Up Measurement Parameters
The configuration tab is used to set up your LCR Pro1 measurement parameters. Figure 5-1 shows a typicalscreen in the configuration tab.

Figure 5-1: Configuration Tab

NOTE(1). The hardware reset function can reset all the device parameters to factory original settings no matter whichmode it is currently in. To perform hardware reset, just press down both "Up" and "Down" buttons on the device
for more than 10 seconds until the device display is off. Releasing both buttons will reboot the device.
(2). After the device is reset, all the self-calibration data will be removed. Please perform self-calibration again if
necessary.

Measurement Setup
The Autoset button is used to reset all the measurement parameters to default. Once it is clicked, the following
parameters are reset to default:
•
•
•
•
•
•
14

LCRZ: Auto
Secondary Parameters: Rs/Rp
Test Frequency: Auto
Test Voltage: 1.0Vrms
Circuit Mode: Auto
Phase Angle: 10°
LCR Data Logger User Manual - ver 1.04

The Measurement dropdown menu provides the following options:• LCRZ: Measure inductance (L), capacitance (C), resistance (R) or impedance (Z).
• ESR: Measure the equivalent series resistance (ESR) of a capacitor, independent of its capacitance.
• DCR: Measure the resistance of an unknown component by direct current resistance (DCR) mode.
• Diode: Test diode polarity or indicate a faulty diode if it sees short/open conditions.
• Sorting: Sort the component according to the preset tolerance.
• Recording: Record the maximum, minimum, average result and measurement cycles during a series ofmeasurements.
• Continuity: Test whether the component under test is electrically connected.
The Primary Setting dropdown menu provides the following options:• Auto: the meter automatically identifies the component under test (L, C or R).
• R: measure resistance.
• L: measure inductance.
• C: measure capacitance.
• Z: measure impedance.
The Sec. Setting (Secondary Setting) dropdown menu provides the following options:• Rs/Rp: the meter displays series resistance (Rs) or parallel resistance (Rp). If the circuit mode is selected
as series, Rs will be displayed. If the circuit mode is selected as parallel, Rp will be displayed.
• D/Q/θ: the meter displays dissipation factor (D), quality factor (Q) or phase angle (θ). If a capacitor isbeing measured, D will be displayed. If an inductor is being measured, Q will be displayed. If an resistor isbeing measured, θ will be displayed.
The Test Frequency dropdown menu provides the following options:• Auto: auto frequency (AF) mode, the meter automatically selects the best test frequency.
• 100Hz: set test frequency to 100Hz.
• 120Hz: set test frequency to 120Hz.
• 1kHz: set test frequency to 1kHz.
• 10kHz: set test frequency to 10kHz.
• 100kHz: set test frequency to 100kHz.
The Test Voltage dropdown menu provides the following options:• 1.0Vrms: set test voltage to 1.0Vrms.
• 0.5Vrms: set test voltage to 0.5Vrms.
• 0.2Vrms: set test voltage to 0.2Vrms.
The Phase Angle dropdown menu provides the following options:
• 5°
• 10°
• 15°
• 20°
• 30°
• 45°

NOTE
15

LCR Data Logger User Manual - ver 1.04

Phase angle menu is used to set the phase angle condition for auto component identification. When the meterworks in auto mode, it automatically identifies the component type according to the phase angle setting.

16

LCR Data Logger User Manual - ver 1.04

The Continuity Thr. (Continuity Threshold) dropdown menu provides the following options:• 0.1Ω• 1Ω• 10Ω• 100Ω• 1kΩ
The Circuit Mode dropdown menu provides the following options:• Auto: set to auto circuit mode. Series or parallel circuit mode will be automatically identified.
• Series: set to series circuit mode.
• Parallel: set to parallel circuit mode.

Sorting Setup
When sorting mode is selected, the options in sorting setup section will become active. Figure 5-2 shows atypical sorting screen.

Figure 5-2: Sorting Screen with Passed Result
The Tolerance dropdown menu provides the following options:• 0.5%• 1.0%• 5.0%• 10.0%• 20.0%• 30.0%17

LCR Data Logger User Manual - ver 1.04

The Reference shows the current reference value for sorting.
Ref. Value R/L/C allows you to input a new reference value. Once the Save button is clicked, the new value will
overwrite the old reference value.
In sorting mode, the LCR Data Logger gets the measurement result from the device and calculates the error
percentage using the measured result and the reference value. Then it compares the error percentage with thetolerance range. If the result is within the range, a green pass message will be on the display as Figure 5-2
shows. If the result is out of range, a red fail message will be on the display as Figure 5-3 shows.

Figure 5-3: Sorting Screen with Failed Result

18

LCR Data Logger User Manual - ver 1.04

Setting Up System Parameters
The system tab is used to set up your LCR Pro1 system parameters. Figure 6-1 shows a typical screen in thesystem tab.

Figure 6-1: System Tab

System Setup
The System Autoset button is used to reset all the system parameters to default. Once it is clicked, the following
parameters are reset to default:
•

•
•
•
•

Display: Right Hand
Contrast: 40%
Sound: On
Timeout: 1 Minute
Test Speed: 1.0 Second

The Turn Display Off button is used to turn off the display. Once the display is turned off, the message on the
button is changed to Turn Display On. Click it again to turn the display back on.
The Display dropdown menu provides the following options:• Right Hand: select the right hand operation mode.
• Left Hand: select the left hand operation mode.

19

LCR Data Logger User Manual - ver 1.04

The Contrast option is used to set display brightness. It provides the following options:
• 20%
• 30%
• 40%
• 50%
• 60%
• 70%
• 80%
• 90%
• 100%The Sound dropdown menu provides the following options:• Sound On: turn on sound.
• Sound Off: turn off sound.
The Timeout option is used to select the timer to powers off the meter automatically if neither a measurement
is performed nor any button is clicked. It provides the following options:• 30 Seconds• 1 Minute• 2 Minutes• 4 Minutes

NOTE
The timeout setting only works when the LCR Pro1 is in standalone mode. When it is connected to PC, its display
will be always on unless the "Turn Display Off" button is clicked.
The Test Speed option is used to select how fast the meter measures the component under test. It provides thefollowing options:• 0.5 Second• 1.0 Second• 2.0 Second

Device Information
Once a device is connected to PC, this section shows the device hardware version, software version and device
model.

Self-Calibration
The self-calibration includes short and open calibration. It offsets the meter’s internal and external parasitic
parameters for better accuracy. Performing this action will help the user correct the influence for temporary uses.
Click the Start button beside the Open Calibration message to start open calibration.
Click the Start button beside the Short Calibration message to start short calibration.

20

LCR Data Logger User Manual - ver 1.04

GUI
It shows the version of the LCR Data Logger.

21

LCR Data Logger User Manual - ver 1.04

Using Multiple Component Identification Function
You can perform multiple component identification testing on the multipleID tab. In multiple component
identification testing, the LCR Data Logger software configures the device to do step by step measurement on
several components. You don’t need to switch parameters manually when switching from one component toanother. The device automatically sets the parameters according to the programming. It records the results andreports Pass or Fail.
Figure 7-1 shows a typical screen in the multipleID tab.

Figure 7-1: MultipleID Tab

Parameters Setup
When you enter the multipleID tab, only the Create Profile and Open Profile buttons are active. Click theCreate Profile button to create a new profile and click the Open Profile button to open an existing profile.
Once the Create Profile button is clicked, a dialog box prompts as Figure 7-2 shows. Select Yes to create avisual profile and select No to create a non-visual profile.

NOTE
In visual profile, you can load the picture of the board under testing and use it to indicate different test points.
Please refer to the "Visual Multi-Component Measurement" section on page 22 for details.

22

LCR Data Logger User Manual - ver 1.04

Figure 7-2: Create a Profile
The following grayed out options also become active for you to program the parameters for each component
going to be tested.
The Primary Setting dropdown menu provides the following options:• R: measure resistance.
• L: measure inductance.
• C: measure capacitance.
• Z: measure impedance.
The Component box is used to add a reference for the component under testing. Such reference name and/or
number will be recorded in the table with other testing parameters and results during multiple component
identification testing period.
The Test Frequency dropdown menu provides the following options:• Auto: auto frequency (AF) mode, the meter automatically selects the best test frequency.
• 100Hz: set test frequency to 100Hz.
• 120Hz: set test frequency to 120Hz.
• 1kHz: set test frequency to 1kHz.
• 10kHz: set test frequency to 10kHz.
• 100kHz: set test frequency to 100kHz.
The Test Voltage dropdown menu provides the following options:• 1.0Vrms: set test voltage to 1.0Vrms.
• 0.5Vrms: set test voltage to 0.5Vrms.
• 0.2Vrms: set test voltage to 0.2Vrms.
Input Ref. R/L/C Value allows you to input a reference value for the software to compare with the actual23

LCR Data Logger User Manual - ver 1.04

measurement result. You can select the unit as well from the dropdown menu on its right side.
The Pass/Fail Criteria dropdown menu provides the following options:• 0.1%• 0.2%• 0.5%• 1.0%• 5.0%• 10.0%• 20.0%• 30.0%
Click the Next Component button to start to program the next component. Click the Prev. Component button
to go back to the previous component. Click the Delete button to delete the current component. Once all theparameters are set up, click the Save Component button to save the current component.
The View Image button is used for visual profile only. Once it is clicked, it switches to the image from the LCR
Data Logger window.
The Lock Profile option is used to prevent the profile from being accidentally modified. The default setting is
unchecked when you create a new profile and checked when you open an existing profile. If it is necessary tomodify an existing profile, please uncheck this option.
After you are done all programming, you can click the Save Profile As button to save the profile. Once it isclicked, a dialog box prompts to ask for the location to save the file as Figure 7-3 shows. The default location is
at Windows (C:) > LCR_DATA.

Figure 7-3: Save a Profile File24

LCR Data Logger User Manual - ver 1.04

The Open Profile button is located on the left side of the Save Profile As button. It is used to open a saved
profile as Figure 7-4 shows.

Figure 7-4: Open a Profile File

Visual Multi-Component Measurement
When you use visual measurement, the LCR Data Logger software programs the device parameters and shows
you on the picture where the component under testing is located. It makes your test more accurately and quickly
as you can easily find the right component on the picture.
To create a visual profile, click the Create Profile button. You will see a dialog box prompts saying that "Create
visual profile from PCB image?” Select the Yes option on that dialog box, then select the picture you want to use.

25

LCR Data Logger User Manual - ver 1.04

A "PCB Graphic Interface" window prompts as Figure 7-5 shows.

Figure 7-5: PCB Graphic Interface Window
The control panel is on the top left side. It provides the following functions:• Accept 1st Test Point: click this button to confirm the location of the 1st test point you draw
• Accept 2nd Test Point: click this button to confirm the location of the 2nd test point you draw
• Zoom In: zoom in the picture• Zoom Out: zoom out the picture• Return to LCR Data Logger: return to the LCR Data Logger window if it is covered by the picture• Stop MI Test: click this button to stop the Multi Component Identification test
• Pause MI Test: click this button to pause the Multi Component Identification test
• Ref. designator: show the designator of the component under testing• R= (or L=, C=, Z=): show the type and value of the component under testing• Test Tolerance: show the tolerance setting of the component under testing• Test Volt/Freq: show the test voltage and test frequency

NOTE(1). It is not necessary to make 2 test points for the components that only have 2 terminals, such as resistor,
capacitor, etc. One test point will be enough.
(2). The 2nd test point option can be used when testing IC different pins or 2 test points that are far away, suchas one test point vs ground.
The whole picture is displayed on the bottom left side. The right side is your working area to edit your picture.
You can select the right size by zooming in or out.

NOTE
26

LCR Data Logger User Manual - ver 1.04

You can drag the picture by pressing down your mouse right key and moving your mouse.
To select the test point, please press down your mouse left key and drag. The selected area will be highlighted inpurple. Figure 7-6 shows a 1KΩ resistor has been selected as a test point.

Test point is selected.

Figure 7-6: Select Test Point
Once the test point is selected, please click the Accept 1st Test Point button to confirm your selection. Once
confirmed, the purple area turns into yellow as Figure7-7 shows. A pair of yellow cursors are displayed on thewhole picture to indicate where this component is located as Figure 7-7 shows.

A pair of yellow
cursors are displayed
to indicate where thecomponent is located.

27

1st Test point isconfirmed.

LCR Data Logger User Manual - ver 1.04

Figure 7-7: Confirm the 1st Test Point
Then go back to the LCR Data Logger window and input component designator and reference value as Figure 7-8 redboxes show. Select the proper test frequency, test voltage and pass/fail criteria as well. After all the parameters are set up,
please save them by clicking the Save Component button as Figure 7-8 green box shows.

Figure 7-8: Configure Component Parameters in Visual Multiple Component Testing
After you are done with the current component, you can start to program the next component by clicking the
Next Component button. If you want to select two test points for the component under testing, please take thefollowing steps:
1.
2.
3.
4.
5.
6.
7.

28

Press down your mouse left key on the picture and drag the area for the 1st test point.
Click the Accept 1st Test Point button to confirm your selection.
Press down your mouse left key on the picture and drag the area for the 2nd test point.
Click the Accept 2nd Test Point button to confirm your selection.
Go to the LCR Data Logger window and input component designator and reference value.
Select the proper test frequency, test voltage and pass/fail criteria.
After all the parameters are set up, please save them by clicking the Save Component button.

LCR Data Logger User Manual - ver 1.04

Figure 7-9 shows the picture that two test points are confirmed. As you can see, the 1st test point is highlighted
in yellow and the 2nd test point is highlighted in orange which matches the background color of thecorresponding buttons.

1st Testpoint inyellow.

2nd Test point
in orange.

Figure 7-9: Confirm Two Test Points
After you are done with all components, you can click the Save Profile As button to save the profile.
To test the whole profile, please click the Test All Components button. It will test all the components one byone. To test only the current component you selected, please click the Test Selected Component button.
Once the Test All Components button is clicked, the LCR Data Logger software uses the parameters in the
profile to set the device for testing the first component. A dialog box prompts to indicate which component isgoing to be tested. On the picture, this component is highlighted.

29

LCR Data Logger User Manual - ver 1.04

Figure 7-10 shows an example for testing a 1KΩ resistor. The component information shows on the left side,
such as designator, type, value and tolerance. The location of this resistor is indicated by a pair of cursors
displayed on the whole picture on the bottom left side.

Indicate which component
to be tested and thecomponent type, value,
tolerance.

Yellow cursors indicate
the component location.

Test point inhighlighted with solid
yellow lines.

Figure 7-10: Start to Test a 1KΩ Resistor in Visual Multiple Component Testing
Find the component under testing and place your device test tips on it to start testing. The real time result
displays as Figure 7-11 shows:

Real time result
during measurement.

Figure 7-11: Test a 1KΩ Resistor in Visual Multiple Component Testing
If the result meets the pass/fail criteria, a green pass message will be on the display. On the picture, the30
LCR Data Logger User Manual - ver 1.04

corresponding test point turns into green as well. If the result doesn't meet the pass/fail criteria, a red failmessage and the red test point will be displayed.
Figure 7-12 shows an example of the 1KΩ resistor test result.

Green PASS
message

Test point inhighlighted in green to
indicate a PASS result.

Figure 7-12: 1KΩ Resistor Test Result in Visual Multiple Component Testing
Next, the LCR Data Logger software sets the device parameters for the next component under testing. A dialog
box prompts again to indicate the user which component is going to be tested next.
Figure 7-13 shows an example for testing a 100uH inductor. As you can see, the prompted dialog box says totest component L1. The location of this inductor (two test points) is indicated by two pair of cursors displayed on
the whole picture on the bottom left side.

31

LCR Data Logger User Manual - ver 1.04

Figure 7-13: Start to Test a 100uH Inductor in Visual Multiple Component Testing
Find the component under testing and place your device test tips on it to start testing. If the result meets thepass/fail criteria, a green pass message will be on the display. On the picture, the corresponding test point turns
into green as well. If the result doesn't meet the pass/fail criteria, a red fail message and the red test point will be
displayed.
Figure 7-14 shows an example of the 100uH inductor test result.

Figure 7-14: 100uH Inductor Test Result in Visual Multiple Component Testing
32

LCR Data Logger User Manual - ver 1.04

Keep testing on each component. When you are done, a dialog box prompts to remind you to save the test
results or keep editing current profile as Figure 7-15 shows. You can save the test results by clicking the Export
button. To keep editing the profile, please click the Clear button. It will keep the profile settings, but all the test
results will be deleted.

Figure 7-15: Finish the Multiple Component Identification Testing
All the test parameters and data are recorded in a spreadsheet. It has a Pass/Fail column to tell you whether the
result is a PASS or FAIL as Figure 7-16 shows.

33

LCR Data Logger User Manual - ver 1.04

Figure 7-16: Test Result Spreadsheet in Visual Multiple Component TestingOnce the Export button is clicked, a dialog box prompts to ask for a location for saving the test results as Figure
7-17 shows. Press the Save button on the dialog box to save the file. The default location is at Windows(C:) >LCR_DATA.

Figure 7-17: Save the Test Results in Visual Multiple Component Testing
After the test results are saved, a dialog box prompts to indicate the file is saved successfully as Figure 7-18
shows.

34

LCR Data Logger User Manual - ver 1.04

Figure 7-18: Successfully Saved the Test Results in Visual Multiple Component Testing

Non-Visual Multi-Component Measurement
The setup procedures of non-visual multiple component measurement are similar to those of visual multiple
component measurement except you don't need to set up test points on the picture.
To create a non-visual profile, click the Create Profile button. You will see a dialog box prompts saying that
"Create visual profile from PCB image?” Select the No option on that dialog box.
In LCR Data Logger window, input component designator and reference value as Figure 7-19 red boxes show.
Select the proper test frequency, test voltage and pass/fail criteria as well. After all the parameters are set up,
please save them by clicking the Save Component button as Figure 7-18 green box shows.

35

LCR Data Logger User Manual - ver 1.04

Figure 7-19: Configure Component Parameters in Non-Visual Multiple Component Testing
After you are done with the current component, you can start to program the next component by clicking the
Next Component button.
After you are done with all components, you can click the Save Profile As button to save the profile.
To test the whole profile, please click the Test All Components button. It will test all the components one byone. To test only the current component you selected, please click the Test Selected Component button.
Once the Test All Components button is clicked, the LCR Data Logger software uses the parameters in the
profile to set the device for testing the first component. A dialog box prompts to indicate which component isgoing to be tested.

36

LCR Data Logger User Manual - ver 1.04

Figure 7-20 shows an example for testing a 1KΩ resistor. As you can see, the prompted dialog box says to test
component R1.

Figure 7-20: Start to Test a 1KΩ Resistor in Non-Visual Multiple Component Testing
Find the component under testing and place your device test tips on it to start testing. If the result meets thepass/fail criteria, a green pass message will be on the display. Otherwise, a red fail message will be displayed.
Figure 7-21 shows an example of the 1KΩ resistor test result.

37

LCR Data Logger User Manual - ver 1.04

Figure 7-21: 1KΩ Resistor Test Result in Non-Visual Multiple Component Testing
Next, the LCR Data Logger software sets the device parameters for the next component under testing. A dialog
box prompts again to indicate the user which component is going to be tested next.
Keep testing on each component. When you are done, a dialog box prompts to remind you to save the test
results or keep editing current profile as Figure 7-22 shows. You can save the test results by clicking the Export
button. To keep editing the profile, please click the Clear button. It will keep the profile settings, but all the test
results will be deleted.

Figure 7-22: Test Result Spreadsheet in Non-Visual Multiple Component Testing
All the test parameters and data are recorded in a spreadsheet. It has a Pass/Fail column to tell you the result is a
PASS or FAIL as Figure 7-23 shows.

38

LCR Data Logger User Manual - ver 1.04

Figure 7-23: Test Result Spreadsheet in Non-Visual Multiple Component TestingOnce the Export button is clicked, a dialog box prompts to ask for a location for saving the test results as in
Figure 7-24. Press the Save button on the dialog box to save the file. The default location is at Windows(C:) >LCR_DATA.

Figure 7-24: Save the Test Results in Non-Visual Multiple Component Testing
After the test results are saved, a dialog box prompts to indicate the file is saved successfully as in Figure 7-25.

39

LCR Data Logger User Manual - ver 1.04

Figure 7-25: Successfully Saved the Test Results in Non-Visual Multiple Component Testing

40

LCR Data Logger User Manual - ver 1.04

Uninstalling Software
The LCR Data Logger software can be removed by taking the following steps:
1. Open the Windows Control Panel window.
2. Select the Uninstall a program option as Figure 8-1 shows.

Figure 8-1: Select Uninstall a Program
3. Browse the list of currently installed programs and find the LCR Data Logger as Figure 8-2 shows.

Figure 8-2: Uninstall LCR Data Logger Software
4. Select the Uninstall button (the red circle in Figure 8-2) to uninstall the LCR Data Logger Software.
5. A dialog box will appear as Figure 8-3 shows. It prompts you to confirm the complete removal of theapplication. Click Yes to proceed. To cancel the uninstallation, click No.

Figure 8-3: Confirm Uninstallation
41

LCR Data Logger User Manual - ver 1.04

6. Follow the instructions on the screen to complete the removal process.

NOTE(1). The above procedures are based on Windows 10. Windows 7 or 8 might have slightly different procedures touninstall a program.
(2). When you update a new version of LCR Data Logger, you may need to uninstall the old version prior to
installing the new version.

42

LCR Data Logger User Manual - ver 1.04