Digital Blood Pressure Monitor
Model UA-767S-W
Instruction Manual Original

EnglishManuel d’instructions Traduction

Français

Manual de Instrucciones Traducción

EspañolManuale di Istruzioni Traduzione

Italiano

Manual de instruções Tradução

Português

Instructiehandleiding Vertaling

Nederlands

Betjeningsvejledning Oversættelse Dansk
Bruksanvisning Översättning

Svenska

Bruksanvisning Oversettelse

Norsk

Ohjekirja Käännös

1WMPD4002904E

Suomi

Talimat Kılavuzu ÇeviriTÜRKÇE‫ﺗﺭﺟﻣﻪ ﺩﻟﻳﻝ ﺍﻹﺳﺗﻌﻣﺎﻝ‬‫ﺗﺮﺟﻤﻪ ﺩﻓﺘﺮﭼﻪ ﺭﺍﻫﻨﻤﺎ‬‫ﻋﺭﺑﻲ‬‫ﻓﺎﺭﺳﯽ‬Contents
Dear Customers .................................................................... 2
Preliminary Remarks ............................................................. 2
Precautions ........................................................................... 2
Parts Identification ................................................................ 5
Symbols ................................................................................ 6
Operation Mode .................................................................... 8
Using the Monitor ................................................................. 9
Installing / Changing the Batteries ................................... 9Connecting the Air Hose ................................................ 10
Connecting the AC Adapter ............................................ 10
Selecting the Correct Cuff Size ....................................... 11
Applying the Arm Cuff ................................................... 11
How to Take Accurate Measurements ............................. 13
Measurement ................................................................. 13
After Measurement ........................................................ 13
Measurements .................................................................... 14
Normal Measurement ..................................................... 14
Measurement with the Desired Systolic Pressure ............ 15
Notes for Accurate Measurement ................................... 15
Recalling the Memory Data ................................................. 16
What Is The IHB/AFib Indicator? .......................................... 17
What Is The AFib? ............................................................... 17％IHB/AFib .......................................................................... 17
Pressure Bar Indicator ......................................................... 18
WHO Classification Indicator ............................................... 18
About Blood Pressure.......................................................... 18
What is Blood Pressure? ................................................. 18
What is Hypertension and How is it Controlled? .............. 19
Why Measure Blood Pressure at Home?........................... 19
WHO Blood Pressure Classification ................................. 19
Blood Pressure Variations............................................... 19
Troubleshooting ................................................................. 20
Maintenance ....................................................................... 21
Technical Data .................................................................... 21

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure
monitor. Designed for ease of use and accuracy, this device will facilitate
your daily blood pressure regimen.
We recommend that you read through this manual carefully before
using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity. (0123:
The reference number to the involved notified body)
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in thehome healthcare environment.
This device is designed to measure blood pressure and pulse rate ofpeople for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremesin temperature, humidity, direct sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff or storing the hose tightly twisted for long
periods, as such treatment may shorten the life of the components.
Take care to avoid accidental strangulation of babies or infants with the hoseand cable.
Do not twist the air hose during measurement. This may cause injury due
to continuous cuff pressure.
The device and cuff are not water resistant. Prevent rain, sweat and waterfrom soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
Wireless communication devices, such as home networking devices, mobile
phones, cordless phones and their base stations, walkie-talkies can affectthis blood pressure monitor.
Therefore, a minimum distance of 30 cm should be kept from such devices.
When reusing the device, confirm that the device is clean.
English 2

Used equipment, parts and batteries are not treated as ordinary
household waste, and must be disposed of according to the applicable
local regulations.
When the AC adapter is used, make sure that the AC adapter can bereadily removed from the electrical outlet when necessary.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the arm must be squeezed by the cuff hardenough to temporarily stop blood flow through the artery. This may cause
pain, numbness or a temporary red mark to the arm. This condition willappear especially when measurement is repeated successively. Any pain,
numbness, or red marks will disappear with time.
Measuring blood pressure too frequently may cause harm due to blood flowinterference. Check that the operation of the device does not result in
prolonged impairment of blood circulation, when using the device
repeatedly.
If you have had a mastectomy, please consult a doctor before using the device.
Do not let children use the device by themselves and do not use the device
in a place within the reach of infants. It may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
Unplug the AC adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially causeburns.
Allow the device to adapt to the surrounding environment before use (about
one hour).
Clinical testing has not been conducted on newborn infants and pregnant
woman. Do not use on newborn infants or pregnant woman.
Do not touch the batteries, the DC jack, and the patient at the same time.
That may result in electrical shock.
Do not inflate without wrapping the cuff around the upper arm.

Contraindications
The following are precautions for proper use of the device.
Do not apply the cuff on an arm with another medical electrical equipment
attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult adoctor before using the device, to avoid medical problems.
Do not self-diagnose the measurement results and start treatment by
yourself. Always consult your doctor for evaluation of the results andtreatment.
Do not apply the cuff on an arm with an unhealed wound.
Do not apply the cuff on an arm receiving an intravenous drip or bloodtransfusion. It may cause injury or accidents.
English 3

Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, suchas a high-pressure oxygen chamber or an oxygen tent. It may cause a fireor explosion.

English 4

Parts Identification
Display

Air Connector Plug

AC adapter Plug
DC Jack
START Button

Air Socket

Air Hose

Proper Fit Range

ArteryIndex MarkArm Cuff
Battery Compartment
Battery Cover

1.5V Batteries(R6P, LR6 or AA)

Display

Average
MEMORYIHB/AFib symbol
Systolic
Pressure%IHB/AFib
Cuff Fit Error Symbol

Diastolic
Pressure

Movement Error Symbol
Heart MarkWHO Classification Indicatorand Pressure Bar Indicator

Pulse Rate

Battery Indicator
English 5

Symbols
Symbols that are printed on the device caseSymbols
SYS
DIA
PUL

IP

Function / MeaningStandby and Turn the device on.
Systolic blood pressure in mmHg
Diastolic blood pressure in mmHg
Pulse per minuteBattery installation guide
Direct current
Type BF: Device, cuff and tubing are designed to provide
special protection against electrical shocks.
EC directive medical device label
EU-representative
Manufacturer
Date of manufactureInternational protection symbol
WEEE label

SN

Serial number
Refer to instruction manual/booklet
Polarity of DC jack
Keep drySymbols that appear on the display
Symbols

Function / Meaning
Recommended Action
Appears while measurement is in
Measurement is inprogress. It blinks when the pulse isprogress. Remain as
detected.
still as possible.
IHB/AFib symbol
Appears when an irregular heartbeat is
detected.
It may light when a very slight vibration
like shivering or shaking is detected.
The reading may yield
an incorrect value. Try
Appears when a body or arm movement
the measurement again.
is detected.
Remain still during
measurement.

English 6

Symbols
Symbols

Function / Meaning

Recommended Action
The reading may yield an
Appears during measurement when incorrect value. Apply thethe cuff is attached loosely
cuff correctly, and try themeasurement again.
Detected rate of IHB/AFib in memory%IHB/AFib =Number of detected
IHB/AFibs in memory
Total number

x100 [%]

Previous measurements stored in
MEMORY.
Average data
FULL BATTERY
The battery power indicator during
measurement.
LOW BATTERY
The battery is low when it blinks.
Unstable blood pressure due tomovement during measurement.

Replace all batteries with newones when the mark blinks.
Try the measurement again.
Remain very still during
measurement.

The systolic and diastolic values
are within 10 mmHg of each other.
The pressure value did not increase
Apply the cuff correctly,
during the inflation.
and try the measurement
again.
The cuff is not applied correctly.
PUL DISPLAY ERROR
The pulse is not detected correctly.
Blood pressure monitor internal
errorEnglish 7

Remove the batteries andpress the START button,
and then install the batteries
again. If the error still
appears, contact the dealer.

Operation Mode
1. Normal Measurement
Press the START button. Blood pressure is measured and the data is stored
in memory. This device can store the last 60 measurements in memory.

2. Recalling the Data
When nothing is displayed, press and holdthe START button.

In standby,press and hold the
STARTbuttonRelease the button when displaying the
average data.

The data number and stored data areautomatically displayed in order from the lastmeasurement.

For details on recalling the data, Refer to“Recalling the Memory Data”.

3. Deleting all Data Stored in Memory
When turning off the device, press and hold the
START button until the "" is displayed.
Select "" to clear data.
Data is cleared when themark blinks.
The device turns off automatically.

START

4. Measurement with the Desired Systolic Pressure
Refer to page 15 for measurement with the desired systolic pressure.

English 8

Using the Monitor
Installing / Changing the Batteries
1. Remove the battery cover.

Step 1
Step 3

2. Remove the used batteries andinsert new batteries into the battery
compartment as shown, taking carethat the polarities (+ and -) arecorrect.
Use only R6P, LR6 or AA batteries.

Step 1
Step 2

3. Attach the battery cover.

CAUTION
Insert the batteries as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(LOW BATTERY mark) blinks on the display, replace all batteries
with new ones. Do not mix old and new batteries. It may shorten thebattery life, or cause the device to malfunction.
Replace the batteries two seconds or more after the device turns off.
If
(LOW BATTERY mark) appears even after the batteries are replaced,make a blood pressure measurement. The device may then recognize thenew batteries.
(LOW BATTERY mark) does not appear when the batteries are drained.
The battery life varies with the ambient temperature and may be shorter at
low temperatures. Generally, four new LR6 batteries will last approximately
for one year when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device arefor testing monitor performance and may have a limited life.
Remove the batteries if the device is not to be used for a long time.
The batteries may leak and cause a malfunction.

English 9

Using the Monitor
Connecting the Air Hose

Air socket

Insert the air connector plug into
the air socket firmly.

Air connector plug

Connecting the AC Adapter
Insert the AC adapter plug into the
DC jack.
Next, connect the AC adapter toan electrical outlet.

DC jack

AC adapter plug
Use the specified AC adapter.
(Refer to page 21.)
When disconnecting the AC adapter from the electrical outlet, grasp
and pull the AC adapter body out of the outlet.
When disconnecting the AC adapter plug from the blood pressure
monitor, grasp and pull the AC adapter plug out of the monitor.

English 10

Using the Monitor
Selecting the Correct Cuff Size
Using the correct cuff size is important for an accurate reading. If the cuff is
not the proper size, the reading may yield an incorrect blood pressure value.
The arm size is printed on each cuff.
The index
and proper fit range, on the cuff, tell you if you are applying thecorrect cuff. (Refer to "Symbols that are printed on the cuff" on the next page)
If the index
points outside of the range, contact your local dealer topurchase a replacement cuff.
The arm cuff is a consumable. If it becomes worn, purchase a new one.
Arm Size
Recommended Cuff Size
Catalog Number
31 cm to 45 cm
Large Adult Cuff
CUF-F-LA
22 cm to 42 cm
Wide Range Cuff
CUF-I
22 cm to 32 cm
Adult Cuff
CUF-F-A
Arm size: The circumference at the biceps.
Note:
The UA-767S-W is not designed for using a small cuff.

Applying the Arm Cuff
1. Wrap the cuff around the upper arm,
about 1 - 2 cm above the inside ofthe elbow, as shown.
Place the cuff directly against theskin, as clothing may cause a faintpulse, and result in a measurement
error.

Do not roll up shirtsleeve tightly.
Cuff

1 - 2 cm
Air Hose

Artery position mark2. Constriction of the upper arm, causedby tightly rolling up a shirtsleeve, may
prevent accurate readings.

Secure the fabricfastener firmly.

3. Confirm that the index
pointswithin the proper fit range.
Index

English 11

and proper fit range

Using the Monitor
Symbols that are printed on the cuff
Symbols

Function / Meaning

Recommended Action
mark on the
Set theartery of the upper arm orin line with the ring finger
on the inside of the arm.

Artery Position Mark
▲
REF

Index
Catalog number
Proper fit range for the adult cuff.
It's printed on the adult cuff.
Over range printed on the adult cuff
and wide range cuff.

Use the large adult cuff
instead of the adult cuff orwide range cuff.

Proper fit range for the wide range cuff.
It's printed on the wide range cuff.
Proper fit range for the large adult cuff.
It's printed on the large adult cuff.
Under range printed on the adult cuff
and wide range cuff.
Under range printed on the large adult Use the adult cuff instead
cuff.
of the large adult cuff.
Lot number
Large adult cuff

A

Proper fit range

L

Wide range cuff

S

W

L

S

A

L

Adult cuff

English 12

Using the Monitor
How to Take Accurate Measurements
For the most accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross yourlegs. Keep your feet flat on the floor and straighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same level as your heart.
Remain still and keep quiet during measurement.
Do not measure right after physical exercise or a bath. Rest for twenty or
thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

Measurement
During measurement, it is normal for the cuff to feel very tight. (Do not be
alarmed).

After Measurement
After measurement, press the START button to turn the device off.
Remove the cuff and record your data.
Note: The device has an automatic power shut-off function, which turns thedevice off approximately one minute after measurement.
Allow at least three minutes between measurements on the same person.

English 13

Measurements
Before measurement, refer to “Notes for Accurate Measurement” on thenext page.

Normal Measurement
1.
2.

3.

4.

5.

Place the cuff on the arm (preferably the left arm).
Sit quietly during measurement.

At heart level

Press the START button.
All of the display segments are displayed.
Press
Zero (0) is displayed blinking briefly.
The display changes, as indicated in the
All of the display
figure at the right, as the measurement
segments
displayed
begins. The cuff starts to inflate. It is
normal for the cuff to feel very tight. A
pressure bar indicator is displayed, on theleft edge of the display, during the inflation.
Zero display
Note: If you wish to stop inflation at any time,
Starts inflationpress the START button again.
When inflation is complete, deflation starts
automatically and(heart mark) blinks,
indicating that the measurement is inprogress. Once the pulse is detected, themark blinks with each pulse beat.
Note: If an appropriate pressure is not obtained,
the device starts to inflate again
automatically.
To avoid re-inflation, refer to “Measurementwith the Desired Systolic Pressure” on thenext page.
When the measurement is complete, thesystolic and diastolic pressure readings
and pulse rate are displayed.
The cuff exhausts the remaining air anddeflates completely.
Press the START button to turn the device off.
After one minute of non-operation, thedevice will turn off automatically.

Note: Allow at least three minutes between
measurements on the same person.
English 14

START

Pressurizing

Measurementin progress
Systolic pressure
Diastolic pressure
WHO classification
Pulse rate
Exhausts remaining
air automaticallyMeasurements
Measurement with the Desired Systolic Pressure
The UA-767S-W is designed to detect the pulse and to inflate the cuff to asystolic pressure level automatically.
Use this method when re-inflation occurs repeatedly or when the results
are not displayed even if the pressure decreases to 20 mmHg or less.
1.

Place the cuff on the arm at heart level(preferably the left arm).

At heart level2.

Press the START button.

Press the START
button3.

When the zero blinks, press and holdthe START button until a number
about 30 to 40 mmHg higher thanyour expected systolic pressure
appears.

4.

When the desired number is reached,release the START button to startmeasurement. Continue to measure
your blood pressure as described on
the previous page.

Release the button
at the desired
systolic pressure

Press and holdthe button

START

Refer to the previous pagefor measurement

Notes for Accurate Measurement

Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five to ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.
Should the device detect a condition that is abnormal, it will stop the measurementand display an error symbol. Refer to page 7 for the description of symbols.
English 15

This device is intended for use by adults. Consult with your physician
before using this device on a child. A child should not use this device
unattended.
The automatic blood pressure monitor’s performance may be affected byexcessive temperature or humidity, or altitude.

Recalling the Memory Data
Note: This device stores the last 60 measurements in memory.
1.

When nothing is displayed, press
and hold the _START_ button torecall the stored data.

Press and START button
hold
Average systolic
Average diastolic

2.

Release the button when displaying
the average data.

3.

The data number and stored data
are automatically displayed in orderfrom the last measurement.

4.

The display will turn off automaticallyafter all data is displayed.

Average pulse
Average of
all the data

Systolic
Diastolic
Pulse
Most recent data

Note: If you press the _START_ button
while recalling data, the device
turns off.

Systolic
Diastolic
Pulse
Last data (Oldest)
English 16

What Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements,
the IHB/AFib indicator will appear on the display with the measurement
values.
Note: We recommend contacting your physician if you see thisIHB/AFib indicator frequently.

What Is The AFib?
The heart contracts due to electrical signals occurring in heart and sends
blood through the body. Atrial fibrillation (AFib) occurs when the electricalsignal in the atrium becomes confused and leads to disturbances in thepulse interval. AFib can cause blood to stagnate in the heart, which caneasily create clots of blood, a cause of stroke and heart attack.

％IHB/AFib%IHB/AFib is displayed as frequency of IHB/AFib detected.
IHB/AFib can detect not only noises such as physical movement but also
an irregular heartbeat. Therefore, we recommend contacting yourphysician if %IHB/AFib level is high.
Number of detected
IHB/AFibs in memory
x 100 [%]
%IHB/AFib =Total number
Display of %IHB/AFib: %IHB/AFib is displayed when displaying average values.
(Refer to “2. Recalling the Data” in “Operation Mode”)%IHB/AFib is not displayed when the memory number is six or less.
Average value display

Level 0
%IHB/AFib=0

Level 1
%IHB/AFib=1 - 9

Level 2
%IHB/AFib=10 - 24

Not displayed

English 17

Level 3
%IHB/AFib=25 - 100

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing

Releasing air

Inflation in progress

Measurement in progress
Inflation complete

WHO Classification Indicator
Each segment of the bar indicator
corresponds to the WHO blood pressureclassification described on the next page.
WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimal: The indicator displays a segment, basedon the current data, corresponding to the
WHO classification.

Example:Moderate hypertension

Mild hypertension

High normal

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of the arteries.
Systolic pressure occurs when the heart contracts. Diastolic pressure
occurs when the heart expands. Blood pressure is measured in millimetersof mercury (mmHg). One's natural blood pressure is represented by thefundamental pressure, which is measured first thing in the morning while
one is still at rest and before eating.
English 18

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left unattended,
can cause many health problems including stroke and heart attack.
Hypertension can be controlled by altering lifestyle, avoiding stress, andwith medication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?
Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

WHO Blood Pressure Classification

Blood Pressure Variations

Diastolic blood pressureStandards to assess high Reference Material: Journal of Hypertension 1999, Vol 17 No.2
Grade 3 hypertension (severe)
blood pressure, without mmHg
110
regard to age, have been
Grade 2 hypertension (moderate)105
established by the World
100
Health Organization (WHO),
Grade 1 hypertension (mild)95
as shown in the chart.
90
85
80

High-normal

NormalOptimalBlood pressureAn
individual’s
blood
120 130 140 150 160 170 180
Systolic blood pressure
mmHg
pressure varies greatly on adaily and seasonal basis. It may vary by 30 to 50 mmHg due to various
conditions during the day. In hypertensive individuals, variations are evenmore pronounced. Normally, the blood pressure rises while at work or
play and falls to its lowest levels during sleep. So, do not be overly
concerned by the results of one measurement.
Typical fluctuation within a day
Take measurements at the same time every(Measured every five minutes)
day using the procedure described in thisSystolic
manual to get to know your normal bloodpressure. Regular readings give a morecomprehensive blood pressure history. Be
sure to note the date and time whenrecording your blood pressure. Consult yourSleep
Diastolicdoctor to interpret your blood pressure data.
PM

English 19

AM

Time

Troubleshooting
ProblemPossible Reason

Recommended Action
Replace all batteries with newBatteries are drained.
Nothing
ones.
appears on the
Reinstall the batteries withdisplay, evenwhen the device Battery terminals are not negative and positive terminals
in the correct position.
matching those indicated on theis turned on.
battery compartment.
Battery voltage is too low.
(LOW BATTERY mark)The cuff does
Replace all batteries with new
blinks. If the batteries are
not inflate.
ones.
drained completely, themark does not appear.
The cuff is not applied
Apply the cuff correctly.
properly.
You moved your arm or
Make sure you remain very stillbody during measurement. and quiet during measurement.
The device does
Sit comfortably and still.
not measure.
The cuff position is not Place your arm on a table with yourReadings arecorrect.
palm facing upward and the cuff at
too high or too
the same level as your heart.
low.
If you have a very weak or
irregular heartbeat, the device
may have difficulty in determining
your blood pressure.
The value is different
Refer to “Why Measure Bloodfrom that measured at a
Pressure at Home?”.
clinic or doctor’s office.
Other
Remove the batteries. Placethem back properly and try themeasurement again.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this product, as any attempt todo so will make your warranty invalid.

English 20

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem using
the troubleshooting instructions, contact the authorized dealer in your areaor our customer service department. The A&D customer service will
provide technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life.
However it is generally recommended to have the device inspected every
2 years, to ensure proper functioning and accuracy. Please contact theauthorized dealer in your area or A&D for maintenance.

Technical DataType
Measurement method
Measurement range

UA-767S-W
Oscillometric measurement
Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / minute
Measurement accuracy
Pressure: ±3 mmHg
Pulse:±5%
Power supply
4 x 1.5V batteries (R6P, LR6 or AA) or
AC adapter (TB-233C) (Not included)
Number of measurements Approx. 700 times LR6 (alkaline batteries)
Approx. 200 times R6P (manganese batteries)
With pressure value 180 mmHg, room temperature
23 °C.
Classification
Internally powered ME equipment (Supplied bybatteries) / Class II (Supplied by adapter)
Continuous operation mode
Clinical test
According to ISO81060-2：2013
In the clinical validation study, K5 was used on 85
subjects for determination of diastolic blood pressure.
EMD
IEC 60601-1-2: 2014
MemoryLast 60 measurements
Operating conditions+10 to +40 °C / 15 to 85%RH / 800 to 1060 hPa
Transport / Storage conditions -20 to +60 °C / 10 to 95%RH / 700 to 1060 hPa
Dimensions
Approx. 140 [W] x 60 [H] x 105 [D] mm
WeightApprox. 245 g, excluding the batteries
English 21

Ingress protection
Applied part
Useful life

Device:
IP20
Cuff
Type BF
Device:5 years (when used six times a day)Cuff:
2 years (when used six times a day)AC adapter: 5 years (when used six times a day)Accessory AC adapter
The adapter is to connect the device to a power source at home. Pleasecontact your local A&D dealer for purchasing. The AC adapter is required to
be inspected or replaced periodically.
TB-233C
Please contact your local A&D dealer forpurchasing.
The AC adapter is required to be inspected or
replaced periodically.
Symbols that are printed on the AC adapter
Symbols
Function / Meaning
For indoor use only
Class II device
Thermal fuse
Fuse
EC directive device label
EAC certification device label
Polarity of AC adapter plug
Accessories sold separately
Arm Size
Cuff
Catalog Number
Cuff Size
CUF-F-LA
Large adult cuff
31 cm to 45 cm
CUF-I
Wide range cuff
22 cm to 42 cm
CUF-F-A
Adult cuff
22 cm to 32 cm
Arm size: The circumference at the biceps.
AC adapter

Catalog Number
TB-233C

Plug (Outlet type)Type C

Note: Specifications are subject to change without prior notice.
IP classification is the degrees of protection provided by enclosures inaccordance with IEC 60529. This device is protected against solid
foreign objects of 12 mm diameter and greater such as a fingers. This
device is not protected against water.

English 22

Sommaires
Chers clients ......................................................................... 2
Remarques préliminaires ...................................................... 2
Précautions ........................................................................... 2
Nomenclature ....................................................................... 5
Symboles .............................................................................. 6
Mode opératoire ................................................................... 8
Utilisation de l’appareil ......................................................... 9Mise en place / Remplacement des piles .......................... 9
Branchement du tuyau d’air ........................................... 10
Branchement de l’adaptateur secteur ............................. 10
Choix du brassard ......................................................... 11
Mise en place du brassard .............................................. 11
Comment faire des mesures correctes .............................. 13
Pendant la mesure ......................................................... 13
Après la mesure ............................................................. 13
Les mesures ....................................................................... 14
Mesure normale ............................................................. 14
Mesure avec la pression systolique désirée .................... 15
Consignes et observations pour une mesure correcte..... 15
Rappel des données mémorisées ........................................ 16
Qu’est ce que l’indicateur de RCI/FA? ................................. 17
Qu’est-ce que la FA? ........................................................... 17％RCI/FA ............................................................................. 17
Indicateur de progression du gonflage ................................ 18
Indicateur de tension selon la classification OMS................. 18
Pression sanguine ............................................................... 18
Qu'est-ce que la pression sanguine? .............................. 18
Qu'est-ce que l'hypertension et comment la contrôler? .. 19
Comment mesurer la pression sanguine chez soi?.......... 19
Classification de la tension selon l’OMS ......................... 19
Variations de la pression sanguine ................................. 19
Résolution des problèmes ................................................... 20
Entretien ............................................................................. 21
Fiche technique .................................................................. 21

Français

1

Chers clients
Félicitations pour l’achat de ce moniteur de la pression sanguine A&D derniercri. Etant étudié pour faciliter l’utilisation et la précision, ce moniteur vous
donnera au quotidien des mesures précises de la pression sanguine.
Lisez ce manuel entièrement et soigneusement avant d’utiliser l’appareil
pour la première fois.

Remarques préliminaires
Cet appareil est conforme à la directive européenne 93/42 EEC des appareils
de conformité. (0123: Le numéro demédicaux avec le marquage
référence au corps notifié concerné)
Cet appareil est conçu pour être utilisé sur les adultes, et non sur lesnouveau-nés ou les enfants.
Lieu d’utilisation. Cet appareil est destiné à être utilisé par vous-même dans unenvironnement domestique de soins de santé.
Cet appareil est conçu pour mesurer la tension artérielle et la fréquence cardiaquedes personnes dans un objectif de diagnostic.

Précautions
Des composants de précision sont utilisés dans la construction de cet appareil. Il
convient d'éviter les extrêmes en matière de température, d'humidité, d'exposition à
la lumière directe du soleil, de chocs ou de poussière.
Nettoyez l'appareil et le brassard à l'aide d'un chiffon sec et doux ou d'un chiffon
imbibé d'eau et d'un détergent neutre. N'utilisez jamais d'alcool, de benzène, dediluant ou d'autres produits chimiques agressifs pour nettoyer l'appareil ou lebrassard.
Évitez de plier fortement le brassard ou de ranger le tuyau fortement tordu pendant
des périodes prolongées, car cela pourrait réduire la durée de vie des accessoires.
Veillez à ce que les bébés et les enfants en bas âge ne s’étranglent pas avec letuyau et le câble.
Ne tordez pas le tuyau d’air pendant la mesure. La pression continue du brassard
pourrait vous blesser.
L'appareil et le brassard ne sont pas étanches. Évitez que la pluie, la sueur ou del'eau ne souillent l'appareil et le brassard.
Il est possible que les mesures soient altérées si l’appareil est utilisé à proximité detéléviseurs, de fours à micro-ondes, de téléphones cellulaires, de sources derayonnement X ou d’autres appareils émettant de forts champs électriques.
Les dispositifs de communication sans fil, tels que les dispositifs réseau à domicile,les téléphones mobiles, les téléphones sans fil et leurs stations de base, lestalkies-walkies peuvent affecter ce moniteur de pression sanguine. Une distanceminimale de 30 cm doit donc être observée pour ces dispositifs.
Si vous réutilisez l'appareil, vérifiez que celui-ci est propre.
Français2

L'équipement, les pièces et les piles usagées ne doivent pas être traités commedes déchets ménagers ordinaires et doivent être mis au rebut conformément auxrèglementations locales applicables.
Lorsque vous utilisez l'adaptateur secteur, assurez-vous que celui-ci peut êtrefacilement retiré de la prise électrique si nécessaire.
Ne modifiez pas l'appareil. Cela pourrait provoquer des accidents ou endommager
l'appareil.
Pour mesurer la pression artérielle, le bras doit être comprimé de manière
suffisamment forte par le brassard pour arrêter temporairement la circulation
sanguine dans l'artère. Il est possible que cela provoque une douleur, unengourdissement ou une marque rouge temporaire sur le bras. Cet état apparaît enparticulier si la mesure est répétée à plusieurs reprises. La douleur,l'engourdissement ou les marques rouges disparaissent avec le temps.
Une mesure trop fréquente de la pression sanguine peut nuire au flux sanguin. Si
vous utilisez l’appareil à plusieurs reprises, assurez-vous que l’utilisation del’appareil n’entrave pas la circulation sanguine de manière prolongée.
Veuillez consulter un médecin avant d’utiliser cet appareil si vous avez subi une
mastectomie.
Ne laissez pas d’enfants utiliser l’appareil par eux-mêmes et n’utilisez pas l’appareildans un endroit où il se trouve à portée d’enfants en bas âge. Cela pourrait
provoquer des accidents ou des dommages.
L’appareil comporte de petites pièces qui risqueraient d’étouffer les enfants en bas
âge s’ils venaient à les avaler.
Débranchez l’adaptateur secteur lorsqu’il n’est pas utilisé pendant la mesure.
L’utilisation d’accessoires non décrits dans ce manuel peut compromettre la sécurité.
Si la pile est en court-circuit, elle peut devenir chaude et causer potentiellement
des brûlures.
Laissez l’appareil s’adapter à l’environnement ambiant avant utilisation (environ
une heure).
Aucun essai clinique n’a été mené sur les nouveau-nés et les femmes enceintes.
N’utilisez pas l’appareil sur des nouveau-nés ou des femmes enceintes.
Ne touchez pas les piles, la prise CC et le patient en même temps. Cela pourrait
entraîner un choc électrique.
Ne gonflez pas sans avoir enroulé le brassard autour de la partie supérieure du bras.

Contre-indications
Vous trouverez ci-dessous les précautions pour l'utilisation appropriée de cet appareil.
N'appliquez pas le brassard sur un bras auquel un autre équipement médicalélectrique est fixé. L'équipement pourrait ne pas fonctionner correctement.
Les personnes ayant de graves problèmes de circulation au niveau du bras doivent
consulter un médecin avant d'utiliser l'appareil afin d'éviter tout problème médical.
Vous ne devez pas procéder à un autodiagnostic du résultat des mesures etcommencer un traitement de votre propre initiative. Consultez toujours un médecinpour une évaluation des résultats et du traitement.
N'appliquez pas le brassard sur un bras comportant une blessure non soignée.
N'appliquez pas le brassard sur un bras recevant un goutte-à-goutte intraveineux ou
une transfusion sanguine. Cela pourrait provoquer une blessure ou des accidents.
Français3

N'utilisez pas l'appareil en présence de gaz inflammables tels que des gaz
anesthésiques. Cela pourrait provoquer une explosion.
N'utilisez pas l'appareil dans des environnements à forte concentration en oxygène,
tels qu'une chambre à oxygène haute pression ou une tente à oxygène. Cela
pourrait provoquer un incendie ou une explosion.

Français4Nomenclature
Affichage

Fiche del'adaptateur
secteurConnecteur de prise d’air

Connecteur decourant continu

Douille d’air

Plage d’accord
appropriée

Bouton START(mise en marche)
Tuyau d’air

Marque d’index

Marque deposition de l’artère

Logement des piles

Brassard
Couvercle logement des piles

Piles 1,5 V(R6P, LR6 ou AA)

Partie d’Affichage

MoyenneMEMOIRE
Symbole de RCI/FA
Pressionsystolique%RCI/FA
Symbole d’erreur de fixation du brassard

Pression
diastoliqueSymbole d’erreur de mouvement
Symbole du cœurIndicateur de la Classification
suivant l’OMS et Indicateur deprogression du gonflage

Pouls

Indicateur des piles
Français

5Symboles
Symboles imprimés sur l’appareil
Symboles
SYS
DIA
PUL

Fonction / Signification
Pour allumer ou éteindre l’appareil
Pression systolique en mmHg.
Pression diastolique en mmHg.
Pouls: nombre de pulsations cardiaques par minute.

IP

Repères pour installer les piles
Courant continu
T Type BF: L’appareil, le brassard et les tubes isolants ont été
conçus de manière à fournir une protection particulière contre leschocs électriques.
Dispositif Médical conforme à la directive européenne 93/42 CEE
Mandataire Européen
FabricantDate de fabrication
Symbole de protection internationale
Déchets d’Equipements Electriques et Electroniques (DEEE)

SN

Numéro de série
Consultez le manuel/la brochure d'instructions
Polarité du connecteur de courant continu
Maintenir sec

Symboles affichés à l'affichage
SymbolesFonction / Signification
Action recommandée
Cet indicateur apparaît lorsque la mesure
L’appareil est en train deest en cours. Il clignote quand lesmesurer; restez calme.
pulsations cardiaques sont détectées.
Symbole de RCI/FA
Cet indicateur apparaît en cas dedétection d’un rythme cardiaque
irrégulier.
Il est susceptible de s’allumer lors de ladétection d’une très faible vibration telle
qu’un frissonnement ou un tremblement.
La lecture peut être àl’origine d’une valeur
Symbole affiché lors de la détection d'un erronée. Recommencez lamesure. Restez
mouvement du corps ou du bras.
parfaitement tranquillependant la mesure.
Français

6Symboles
SymbolesFonction / Signification
Symbole affiché pendant la mesure si lebrassard n’est pas assez serré.

Action recommandée
La lecture peut être àl’origine d’une valeur
erronée. Attachez lebrassard correctement, etrecommencez la mesure.

Taux détecté de RCI/FA en mémoire
Nombre de RCI/FA
détectés en mémoire
x 100 [%]
%RCI/FA =Nombre total
Les mesures précédentes conservées
dans la MEMOIRE.
Données moyennes
PILE NEUVE
Indicateur de tension des piles pendant lamesure.
Remplacer toutes les piles
PILE FAIBLE
par des neuves quand cet
La tension des piles est faible quand il
indicateur clignote.
clignote.
Recommencez la mesure.
Apparaît en cas de fluctuation de laRestez parfaitement
pression due à un mouvement pendant latranquille pendant lamesure.
mesure.
Apparaît si la difference entre la pression
systolique(tension maxima) et la pression
diastolique(tension minima) est inférieureà 10 mmHg.
Apparaît lorsque la pression n’augmente Attachez le brassard
pas pendant le gonflage du brassard.
correctement, etrecommencez la mesure.
Apparaît lorsque le brassard n’est pas
attaché correctement.
ERREUR AFFICHAGE PRESSION
La pression n’est pas détectéecorrectement.
Retirez les piles et appuyez
sur le bouton START , puis
Erreur interne du moniteur de la pression
remettez les piles en place.
sanguine
Si l’erreur s’affiche encore,contactez votre revendeur.

Français7Mode opératoire
1. Mesure normale avec les données conservées
Appuyez sur le bouton START . La pression sanguine est mesurée et les donnéessont conservées dans l’appareil. Cet appareil peut conserver les 60 dernières
données automatiquement dans la MEMOIRE.

2. Rappel des données en mémoire
Si rien ne s’affiche, appuyez de manière
prolongée sur le bouton START .

En veille,
appuyez de manière
prolongée sur le
START bouton

Relâchez le bouton lorsque les données
moyennes s'affichent.
Le numéro des données et les données
enregistrées s'affichent automatiquement dansl'ordre, à partir de la dernière mesure.
Pour plus de détails sur le rappel des données,
reportez-vous à la section « Rappel des donnéesmémorisées ».

3. Suppression de toutes les données en mémoire
Lorsque vous éteignez l’appareil, appuyez sur le bouton_START_ et maintenez-le enfoncé jusqu'à ce que«» s'affiche.
» pour effacer les données.
Sélectionnez «clignote.
Les données sont effacées lorsque le symbole
L'appareil s'éteint automatiquement.

4. Mesure avec la pression systolique désirée
Voir la page 15 pour la mesure avec la pression systolique désirée.
Français

8

START

Utilisation de l’appareil
Mise en place / Remplacement des piles
1. Enlevez le couvercle du logement despiles.

Etape 1

Etape 3

2. Enlevez des piles usées et insérez despiles neuves dans le logement en
respectant bien les polarités (+ et -)indiquées.
Utilisez uniquement des piles R6P, LR6
ou AA.

Etape 1
Etape 2

3. Fermez le couvercle du logement des piles.

ATTENTION
Veillez à insérer les piles dans le bon sens, sinon l’appareil ne fonctionnera pas.
Quand le symbole(PILE FAIBLE) clignote sur l’écran, l’appareil indique queles piles doivent être remplacées. Changer alors toutes les piles par des neuves.
Ne mélangez pas les piles neuves et anciennes car cela pourrait raccourcir leurdurée de vie ou détériorer l’appareil.
Remplacez les piles au moins deux secondes après avoir éteint l'appareil.
(PILE FAIBLE) s'affiche même après le remplacement desSi le symbole
piles, réalisez une mesure de la pression sanguine. L'appareil devrait alors
reconnaître les nouvelles piles.
Le symbole(PILE FAIBLE) n'apparaît pas lorsque les piles sont épuisées.
La durée de vie des piles varie avec la température ambiante et peut êtreraccourcie par des températures basses. En général, quatre piles neuves LR6
durent approximativement un an en cas de mesure deux fois par jour.
Utilisez uniquement les piles spécifiées. Les piles fournies avec l'appareil sontpour tester les performances de l'écran d’affichage et peuvent donc avoir une
durée de vie plus courte.
Retirez les piles lorsque l'appareil n'est pas utilisé pendant un long moment. Lespiles pourraient couler ce qui entraînerait un mauvais fonctionnement del'appareil.

Français

9

Utilisation de l’appareil
Branchement du tuyau d’air

Douille d'air

Insérez fermement la connecteur deprise d’air dans le douille d’air.

Connecteur de prise d'air

Branchement de l’adaptateur secteur
Introduire la fiche de l'adaptateur secteurdans le connecteur de courant continu.
Brancher ensuite l’adaptateur sur une
prise de courant.

Connecteur de courant
continu

Fiche de l'adaptateur
Utilisez l'adaptateur secteur spécifié.
secteur(Voir la page 21.)
Lorsque vous débranchez l’adaptateur secteur de la prise secteur, tenez lecorps de l’adaptateur et débranchez-le de la prise.
Lorsque vous débranchez l’adaptateur secteur du moniteur de pression
sanguine, tenez la fiche de l’adaptateur et débranchez-la du moniteur.

Français

10

Utilisation de l’appareil
Choix du brassard
L’utilisation d’un brassard adapté à la morphologie du patient est importantepour un résultat précis. Si la taille du brassard n’est pas appropriée il peut en
résulter l’affichage de valeurs erronées.
La taille du brassard (circonférence du biceps) est imprimée sur chaque brassard.
L’indexet la plage d’accord appropriée sur le brassard vous indiquent si
vous mettez le brassard correctement. (Reportez-vous à « Symboles imprimés
sur le brassard », à la page suivante.)
Si l’index
se trouve en dehors de la plage, prière de prendre contact avec
votre revendeur local afin d’acheter le brassard pour le remplacement.
Le brassard est un produit consommable. S’il est usé, achetez-en un neuf.
Taille du bras
31 cm à 45 cm
22 cm à 42 cm
22 cm à 32 cm

Taille du brassard recommandée
Brassard adulte de grande tailleBrassard de grande tailleBrassard adulte

Numéro du catalogue
CUF-F-LA
CUF-I
CUF-F-A

Taille du bras : circonférence au niveau du biceps.
Remarque :le UA-767S-W n'est pas conçu pour une utilisation avec un petit
brassard.

Mise en place du brassard

Ne remontez pas la mancheétroitement.

1. Enroulez le brassard autour du bras,à 1 ou 2 cm au-dessus de l’intérieur ducoude, comme indiqué sur l’illustration
ci-contre. Placez le brassard
directement sur la peau, car unvêtement pourrait assourdir lapulsation et provoquer une erreur demesure.

Brassard
1-2 cmTuyau d’air
Marque de position de l’artère

2. Evitez également de remonterétroitement la manche, car ellecomprimerait le bras ce qui pourrait
aussi provoquer une erreur de mesure.

Veillez à bien fixer
l'attache à tissu.

se trouve à
3. Confirmez que l’index
l’intérieur de la plage d’accord
appropriée.
IndexFrançais

11

et la plage d’accord approriée

Utilisation de l’appareil
Symboles imprimés sur le brassard
SymbolesFonction/Description

Action recommandée
Réglez la marque ● surl’artère du bras supérieur oualignez-la sur l'annulaire surl'intérieur du bras.

Marque de position de l’artère
▲
REF

Index
Numéro du catalogue
Plage d’ajustement appropriée pour lebrassard adulte
Imprimé sur le brassard adulte
Utilisez le brassard adulte dePlage supérieure imprimée sur le brassard grande taille au lieu dubrassard adulte ou du brassard
adulte et le brassard de grande taille.
de grande taille.
Plage appropriée pour le brassard de
grande taille.
Imprimé sur le brassard de grande taille.
Plage d’ajustement appropriée pour lebrassard adulte de grande taille.
Imprimé sur le brassard adulte de grande
taille.
Plage inférieure imprimée sur le brassard
adulte et le brassard de grande taille.
Utilisez le brassard adulte au
Plage inférieure imprimée sur le brassard
lieu du brassard adulte deadulte de grande taille.
grande taille.
Numéro de lot
Plage d’accord appropriée

Brassard adulte
de grande taille

A

L

Brassardde grande taille

S

W

L

Brassard adulte

S

A

L

Français

12

Utilisation de l’appareil
Comment faire des mesures correctes
Pour une meilleure précision des mesures de la pression sanguine :Asseyez-vous confortablement sur une chaise. Posez le bras sur la table. Ne croisez
pas les jambes. Gardez les pieds au sol et tenez le dos bien droit.
Se relaxer pendant 5 à 10 minutes avant de commencer les mesures.
Lever le bras de sorte que le centre du brassard soit au même niveau que le cœur.
Bouger le moins possible et rester calme pendant les mesures.
Ne pas faire de mesure juste après un effort physique ou un bain. Se reposer
pendant 20 à 30 minutes avant de faire les mesures.
S’efforcer de mesurer la tension à la même heure tous les jours.

Pendant la mesure
Pendant la mesure, il est normal que l'on sente que le brassard est très serré.
(Ne pas s'inquiéter.)

Après la mesure
Après la mesure, appuyez sur le bouton START pour éteindre l’appareil.
Enlevez le brassard et notez votre tension et votre pouls.
Note: Cet appareil est doté de la fonction de mise hors tension automatique, qui éteint
un appareil automatiquement dans un délai d’une minute après la mesure.
Attendez au moins trois minutes avant de remesurer votre tension (ou celled’une autre personne) à des fins de confirmation.

Français

13

Les mesures
Avant la mesure, voir «Consignes et observations pour une mesure correcte»
à la page suivante.

Mesure normale
1.

Asseyez-vous confortablement en mettant lebrassard au bras (gauche de préférance).

Au niveau du cœurAppuyez sur le bouton START (mise en marche).
Appuyez START
Tous les segments de l’affichage apparaissent.
Puis, zéro (0) s’affiche en clignotant un instant.
Puis, l’affichage change comme indiqué dans la
Tous les segments del’affichage apparaissent
figure de droite, lorsque la mesure commence.
brièvement.
Le brassard commence à gonfler. Il est normal
que l’on sente que le brassard serre fortement
le bras. Un indicateur de progression du
gonflage est affiché, dans l'angle gauche de lafenêtre, pendant le gonflage.
Zéro s’affiche
Note: Si vous désirez arrêter le gonflage
Gonflage commence
quelque que soit le moment, appuyez denouveau sur le bouton START (miseen marche).
Mise sous
3. Lorsque le brassard est entièrement gonflé, ilpressioncommence à dégonfler automatiquement et le) apparaît
symbole en forme de cœur (pour indiquer que la mesure est en cours.
Une fois le pouls détecté, le symbole clignoteà chaque pulsation.
Note: Si la pression correcte n'est pas obtenue,
l'appareil recommence le gonflage dubrassard automatiquement. Pour éviter unnouveau gonflage, voir la page suivante«Mesure avec la pression systolique désirée».
Mesurage
4. Lorsque les mesures sont finies, les lecturesen cours
des pressions systoliques et diastoliques ainsi
que le nombre de pulsations sont affichées.
Le brassard finit de se dégonfler en chassantPression systolique
l'air restant.
Pression diastolique
5. Appuyer sur le bouton START pour éteindre
Classification suivant l’OMS
l’appareil.
Pouls
L'appareil s’éteint automatiquement après une
minute sans utilisation.
L’air restant est chasséNote: Attendez au moins trois minutes avant deautomatiquement
remesurer votre tension (ou celle d’une autre
personne) à des fins de confirmation.
2.

Français

14

Les mesures
Mesure avec la pression systolique désirée
Le UA-767S-W est étudié pour détecter le pouls et pour gonfler le brassard
jusqu’au niveau de la pression systolique automatiquement.
Utilisez cette méthode si un nouveau gonflage se produit à plusieurs reprises, si
les résultats ne s'affichent pas même si la pression diminue de 20 mmHg ou
moins.
1.

Placez le brassard sur le bras au niveau ducœur (bras gauche de préférence).

2.

Appuyez sur le bouton _START_ .

3.

Lorsque le zéro clignote, appuyez sur lebouton _START_ et maintenez-le enfoncé
jusqu'à ce qu'un nombre de 30 à 40
mmHg supérieur à la pression systoliqueprévue s'affiche.

4.

Relâcher le bouton START pourcommencer la mesure, quand le nombre
désiré est indiqué. Ensuite continuez àmesurer votre tension artérielle commedécrit à la page précédente.

Au niveau du cœurAppuyez sur le START
boutonRelâcher le bouton
quand la pression
systolique désirée est
obtenueAppuyez et maintenez START
enfoncé le bouton

Se référer à la page précédente
pour la mesure.

Consignes et observations pour une mesure correcte

Asseyez-vous confortablement. Posez le bras sur une table, la paume dirigée versle haut et le brassard au même niveau que le cœur.
Détendez-vous pendant cinq à dix minutes avant de commencer la mesure. Si
vous êtes énervé ou au contraire déprimé, cela se reflètera sur votre tension demême que sur votre pouls.
La tension de tout individu varie constamment selon ce qu’il fait et ce qu’il mange.
D’autre part, certaines boissons ont un effet très fort et très rapide sur la tension
artérielle.
Cet appareil se réfère aux pulsations cardiaques pour mesurer la tension et lepouls. Par conséquent, si vos pulsations sont faibles ou irrégulières, l’appareil
pourrait avoir des difficultés à mesurer votre tension artérielle.
Au cas où l’appareil détecte une condition anormale, il s’arrête de mesurer etaffiche un symbole d’erreur. Voir la page 7 pour la description des symboles.
Français

15

Cet appareil est conçu pour être utilisé par des adultes. Avant de l’utiliser pour
mesurer la tension d’un enfant, veuillez en discuter avec votre médecin. Ne
laissez jamais un enfant prendre lui-même sa tension sans surveillance.
La performance du tensiomètre automatique peut être affectée par unetempérature, une humidité ou une altitude excessives.

Rappel des données mémoriséesNote: Cet appareil mémorise les 60 dernières mesures.
Appuyez et maintenez START
enfoncé le bouton

1.

Si rien ne s'affiche, appuyez sur lebouton _START_ et maintenez-leenfoncé pour rappeler les données
enregistrées.

2.

Relâchez le bouton lorsque lesdonnées moyennes s'affichent.

3.

Le numéro des données et les données
enregistrées s'affichentautomatiquement dans l'ordre, à partirde la dernière mesure.

4.

Pression systolique
moyennePression diastolique
moyennePouls moyen
Moyenne de toutes
les données

PressionsystoliquePression
diastolique
PoulsDonnées les plus récentes

L'écran s'éteint automatiquement une
fois toutes les données affichées.

Remarque : Si vous appuyez sur le bouton_START_ lorsque vous rappelez
des données, l'appareil s'éteint.

PressionsystoliquePression
diastolique
PoulsDernières données (les plus anciennes)

Français16

Qu’est ce que l’indicateur de RCI/FA?
Lorsque le moniteur détecte un rythme irrégulier pendant les mesures,l’indicateur de RCI/FA apparaît à l’écran avec les valeurs mesurées.
Note: Nous vous recommandons de contacter votre médecin si vous voyez
fréquemment cet indicateur de RCI/FA :Qu’est-ce que la FA?
Le cœur se contracte en raison des signaux électriques se produisant dans lecœur et envoie le sang dans tout le corps. La fibrillation atriale (FA) se produit
lorsque le signal électrique dans l’oreillette est troublé et provoque desperturbations dans l’intervalle des pulsations. La FA peut causer la stagnation
du sang dans le cœur, ce qui peut facilement créer des caillots sanguins, cause
d’AVC et de crise cardiaque.

％RCI/FA%RCI /FA exprime la fréquence de RCI/FA détecté.
Le RCI/la FA peut détecter non seulement les parasites tels que des mouvements
physiques, mais également un rythme cardiaque irrégulier. Par conséquent, nous
vous recommandons de contacter votre médecin si le niveau du %RCI/FA est élevé.
Nombre de RCI/FA
détectés en mémoire
x 100 [%]
%RCI/FA =Nombre total
Affichage du %RCI/FA: le %RCI/FA apparaît lors de l’affichage des valeursmoyennes. (Reportez-vous à « 2. Rappel des données enmémoire » dans « Mode opératoire »)Le %RCI/FA ne s’affiche pas si le nombre de mémoire est
de six ou moins.
Affichage de la valeur moyenne

Niveau 0%RCI/FA=0

Niveau 1%RCI/FA=1 - 9

Niveau 2
Niveau 3%RCI/FA=10 - 24 %RCI/FA=25 - 100

Non affiché

Français

17

Indicateur de progression du gonflage
L’indicateur montre la progression de la pression pendant la mesure.
Mise sous pression

Dégagement d’air

Gonflage en cours

Mesure en cours
Gonflement complet

Indicateur de tension selon la classification OMS
Chaque segment d’indicateur de progression
de la pression correspond à la classification dela pression sanguine selon l’OMS décrite à lapage suivante.
Indicateur de la classification selon l’OMS
Hypertention sévère
Hypertention modérée
Hypertention faible
Haute-Normale
NormaleOptimale

Exemple:Hypertention modérée

Hypertention faible: L’indicateur affiche un segment, basé surles dernières données, correspondant à laclassification suivant l’OMS.
Haute-Normale

Pression sanguine
Qu'est-ce que la pression sanguine?
La pression sanguine est la force avec laquelle le sang vient frapper contre lesparois des artères. La pression systolique se produit lorsque le cœur se contracte.
La pression diastolique se produit lorsque le cœur se détend.
La pression sanguine est mesurée en millimètre de mercure (mmHg). La pression
sanguine naturelle d’un individu est représentée par la pression fondamentale qui
est mesurée avant toute activité, le matin au lever du lit, et avant de manger ou deboire.
Français18

Qu'est-ce que l'hypertension et comment la contrôler?
L’hypertension est une tension artérielle (ou plus exactement une «pression
sanguine») anormalement élevée. Non surveillée, l’hypertension peut provoquer
de nombreux problèmes de santé, notamment une attaque d’apoplexie ou une
crise cardiaque.
L'hypertension peut être contrôlée en changeant son style de vie, en évitant tout
stress et avec des médicaments pris sous surveillance médicale.
Pour empêcher l’hypertension ou la garder sous contrôle:
Ne pas fumer.
Faire régulièrement des exercices
Réduire sa consommation de sel etphysiques.
Effectuer régulièrement desde graisse.
Maintenir un poids adéquat.
visites médicales.

Comment mesurer la pression sanguine chez soi?
Lorsque la tension est mesurée à l’hôpital ou dans le cabinet du médecin, on neparvient pas toujours à se décontracter, ce qui peut avoir pour effet d’augmenterla tension de 25 à 30 mmHg par rapport à la tension mesurée chez soi. Le fait demesurer sa tension tranquillement chez soi réduit les influences extérieures surla valeur de la tension, complète les mesures effectuées par le médecin etpermet d’établir un relevé plus précis et plus complet de sa tension artérielle.

Classification de la tension selon l’OMS
L’évaluation de l’hypertension,
indépendamment du critère âge, a étéétablie par l’Organisation Mondiale de laSanté (OMS) selon le tableau ci-joint.

Variations de la pression sanguine
La tension artérielle (ou pression sanguine)d’un individu varie fortement selon lemoment de la journée et selon la saison.
Ainsi, elle peut varier de 30 à 50 mmHg
dans une même journée selon un ensemble de conditions. Chez les personnes
hypertendues, ces variations sont souvent encore plus prononcées.
Normalement, la pression augmente pendant le travail ou le jeu et tombe à sonplus bas niveau pendant le sommeil. Par conséquent ne pas trop s'inquiéter
des résultats d'une seule mesure.
Mesurez votre tension tous les jours à lamême heure (en suivant la procédure
décrite dans ce manuel) pour connaître
votre pression sanguine normale. Ces
mesures quotidiennes vous permettrontd’établir un relevé plus complet de votre
tension.
Veillez à bien noter la date et l'heure
lorsque vous enregistrez votre pression.
Consultez votre médecin pour interpréter
les données recueillies.
Français

19

Résolution des problèmes
Problème

Cause probable

Les piles sont usées.
Rien n'apparaît
sur l'affichage,même lorsquel'on met l'appareil Les bornes des piles ne sontpas dans le bon sens.
en marche.

Solution recommandéeRemplacer les piles usées pardes neuves.
Mettre les piles dans le bonsens en faisant correspondre laborne positive et négative
comme indiqué sur le brassard.

La tension des piles est trop
faible.
Remplacer les piles usées par
Le brassard ne se
Le symbole piles faibles
des neuves.
gonfle pas.
apparaît. Si les piles sont usées,ce symbole n’apparaît pas.
Serrer correctement le
Le brassard est mal serré.
brassard.
Vous avez bougé votre bras
Ne pas bouger et rester calme
ou autre pendant la mesure.
pendant la mesure.
L’appareil ne faitS'asseoir confortablement et
pas les mesures.
ne pas bouger. Posez le bras
Les lectures sont Le brassard est mal placé.
sur une table, la paume dirigéetrop hautes ouvers le haut et le brassard au
trop basses.
même niveau que le cœur.
Si votre pouls est faible ou
irrégulier, l'appareil aura desdifficultés à déterminer votre
pression sanguine.
Voir le paragraphe "Comment
La valeur est différente demesurer la pression sanguinecelle mesurée dans unechez soi?".
clinique ou chez le médecin.
AutresRetirer les piles. Le remettre
correctement en place etessayer à nouveau de mesurer.
Note: Si les solutions décrites ci-dessus ne suffisent pas pour résoudre les
problèmes, contactez votre revendeur. N’essayez pas d’ouvrir ou réparervous-même cet appareil car toute tentative invaliderait votre garantie

Français

20

Entretien
Ne pas ouvrir l'appareil car il comporte des composants électriques sensibles et
une unité d'air complexe qui pourraient être endommagés. Si vous n'arrivez pas àsituer un problème avec le tableau de recherche de problèmes, demandez auservice après vente de votre revendeur ou du groupe de service A & D. Le groupede service A & D vous fournira des informations techniques sur les fournisseursagréés, les pièces et les unités de rechange.
Cet appareil a été conçu et fabriqué pour durer. Il est généralement recommandéde le faire vérifier tous les 2 ans pour s’assurer de son fonctionnement correct etde sa précision. Veuillez contacter votre revendeur agréé A&D pour l’entretien.

Fiche techniqueType
Méthode de mesure
Plage de mesure

Précision de mesure
Alimentation électriqueNombre de mesures

Classification

Test clinique

EMD
Mémoire de données
Condition de fonctionnement
Condition de stockage / transport
Dimensions
Poids

UA-767S-W
Mesure oscillométriquePression: de 0 à 299 mmHg
Pression systolique : de 60 à 279 mmHg
Pression diastolique : de 40 à 200 mmHg
Pouls:de 40 à 180 pulsations / minutePression: ±3 mmHg
Pouls:±5%
4 x 1.5V piles (R6P, LR6 ou AA) ou
adaptateur secteur (TB-233C) (Non fournis)
Env. 700 fois LR6 (piles alcalines)
Env. 200 fois R6P (piles au manganèse)
Avec une valeur de pression de 180 mmHg,
température ambiante de 23 °C.
Équipement a alimentation interne ME (fournie parles piles) /Class II (fournie par l'adaptateur)Mode de fonctionnement en continu
Conformément à ISO81060-2：2013
Dans l’étude de validation clinique, K5 a été utilisé
sur 85 sujets pour la détermination de la tension
artérielle diastolique.
IEC 60601-1-2: 2014
Dernières 60 mesuresde +10 à + 40°C / hum. rel. de 15 à 85%/de 800 à 1060 hPa
de -20 à + 60 °C / hum. rel. de 10 à 95 % /de 700 à 1060 hPa
environ 140 [l] x 60 [h] x 105 [p] mmenviron 245 g sans les piles
Français

21

Protection
Partie appliquée
Durée de vie utile

Appareil: IP20
Brassard Type BF
Appareil: 5 ans (six utilisations par jour)Brassard: 2 ans (six utilisations par jour)
Adapteur secteur : 5 ans (lorsqu'utilisé six fois par jour)

Adaptateur secteur accessoire
L'adaptateur permet de brancher l'appareil sur une source d'alimentation domestique.
Veuillez contacter votre distributeur A&D local pour l'acheter. L'adaptateur secteur
doit être périodiquement inspecté ou remplacé.
L’adaptateur secteur doit être inspecté ou remplacé
périodiquement.

Adaptateur secteur
TB-233C

Symboles imprimés sur l’adaptateur secteur
SymbolesFonction / Signification
Utilisation à intérieur uniquement
Appareil de classe II
Fusible thermique
FusibleLabel de conformité de l’appareil à ladirective CE
Label de conformité de l’appareil à lacertification EAC
Polarité de la fiche de l’adaptateur secteur
Accessoires vendus séparément
Brassard Numéro du catalogue
CUF-F-LA
CUF-I
CUF-F-A

Taille du brassard
Brassard adulte de grande
tailleBrassard de grande tailleBrassard adulte

Taille du bras
31 cm à 45 cm
22 cm à 42 cm
22 cm à 32 cm

Taille du bras：La circonférence du biceps
Adaptateur secteur

Numéro du catalogue
TB-233C

Fiche
Type C

Note: Ces caractéristiques techniques sont sujettes à changement sans préavis.
Le classement IP indique les degrés de protection fournis par les coffrets deprotection conformément à la norme CEI 60529. Cet appareil est protégé contre lescorps étrangers solides de 12 mm de diamètre et plus, tel qu’un doigt. Cet appareiln’est pas protégé contre l’eau.

Français22ÍNDICE
Estimados clientes ................................................................ 2
Observaciones Preliminares .................................................. 2
Precauciones ........................................................................ 2
Partes de identificación de componentes ....................................... 5
Símbolos .............................................................................. 6
Modo de operación ............................................................... 8
Uso del monitor .................................................................... 9
Instalación/Cambio de las pilas........................................ 9Conexión de la manguera de aire ................................... 10
Conexión del adaptador de CA ....................................... 10
Selección del brazalete correcto ..................................... 11
Cómo colocar el brazalete .............................................. 11
Para efectuar mediciones correctas ................................ 13
Medición ........................................................................ 13
Después de la medición ................................................. 13
Mediciones ......................................................................... 14
Medición normal ............................................................ 14
Medición con la presión sistólica deseada ...................... 15
Notas para mediciones correctas..................................... 15
Revisión de los datos de la memoria ........................................ 16¿ Qué es el indicador de LI/FibA?........................................... 17¿A qué hace referencia FibA? ............................................... 17％LI/FibA ............................................................................ 17
Indicador de barra de presión ............................................. 18
Indicador de clasificación de la OMS ......................................... 18
Aspectos sobre la tensión arterial ....................................... 18¿Qué es la tensión arterial?............................................. 18¿Qué es la hipertensión y cómo controlarla? ................... 19¿Por qué debe medirse la tensión arterial en la casa? ...... 19
Clasificación de tensión arterial según la OMS ................ 19
Variaciones de tensión arterial ....................................... 19
Localización y corrección de fallos ........................................ 20
Mantenimiento ................................................................... 21
Datos técnicos .................................................................... 21

Español

1

Estimados clientes
FELICIDADES. Usted ha comprado un monitor de presiones sanguíneas A&D, elaparato tecnológico más avanzado y más fácil de usar actualmente en el mercado.
Se recomienda leer este manual de instrucciones detenidamente antes deusarlo por primera vez.

Observaciones Preliminares
Este aparato se ajusta a la Directiva Europea 93/42 EEC para productos médicos.
de conformidad. (0123: Número deEsto se evidencia con la marca
referencia a los quienes notificados)
Este aparato está diseñado para usarse en adultos, los niños y los recién nacidos
no pueden usarlo.
Ámbito de uso. Este aparato es para que pueda usarlo usted mismo en un entorno
doméstico.
Este aparato se ha diseñado para medir la presión sanguínea y la frecuencia delpulso de las personas para el diagnóstico.

Precauciones
En la construcción de este aparato, se han usado componentes de precisión.
Deberán evitarse extremos de temperatura, humedad, rayos solares directos,
golpes o polvo.
Limpie el aparato y el brazalete con un paño seco, suave o un paño humedecido
con agua y un detergente neutro. No use nunca alcohol, gasolina, diluyente u
otros productos químicos para limpiar el aparato o el brazalete.
Evite doblar el brazalete apretado o guardar el tubo retorcido durante períodos
prolongados, dado que ello puede acortar la vida útil de los componentes.
Tenga cuidado para evitar la estrangulación accidental de niños o bebés con eltubo o el cable.
No retuerza la manguera de aire durante la medición. Si lo hace, pueden
producirse lesiones debido a la presión continua del brazalete.
El aparato y el brazalete no son resistentes al agua. Evite que la lluvia, latranspiración y el agua mojen el aparato y el brazalete.
Las mediciones pueden verse afectadas si el aparato se usa cerca detelevisores, hornos microondas, teléfonos móviles, rayos X u otros dispositivos
con campos eléctricos fuertes.
Los aparatos de comunicación inalámbrica, como los dispositivos de red
doméstica, teléfonos móviles, teléfonos inalámbricos y sus centrales ywalkie-talkies, pueden afectar al monitor de presiones sanguíneas. Por lo tanto,
es necesario mantener una distancia de 30 cm entre el monitor y dichos
aparatos.
Español2

Cuando se reutilice el aparato, confirme que está limpio.
El aparato, sus partes y pilas usados no deben considerarse como desperdicios
domésticos ordinarios y deberán eliminarse de acuerdo con los reglamentos
locales aplicables.
Cuando se usa un adaptador CA, asegúrese de que el adaptador CA se puede
retirar rápidamente del enchufe eléctrico cuando sea necesario.
No modifique el aparato. Puede provocar accidentes o daños al aparato.
Para medir la presión sanguínea, se debe apretar el brazo con el brazalete lo
suficiente para detener el flujo de sangre a través de la arteria temporalmente. Esto
puede provocar dolor, hormigueo o una marca roja temporal en el brazo. Este
estado aparecerá especialmente cuando la medición se repite sucesivamente.
Cualquier dolor, hormigueo o marcas rojas desaparecerán con el tiempo.
Si la presión sanguínea se mide con demasiada frecuencia, pueden producirse
daños debido a la interferencia con el flujo sanguíneo. Si utiliza el aparato deforma repetida, compruebe que su uso no provoca una alteración prolongada dela circulación sanguínea.
Si le han practicado una mastectomía, consulte con su médico antes de utilizar
el aparato.
No deje que los niños utilicen el dispositivo sin la supervisión de un adulto, ni
utilice el dispositivo en un lugar dentro del alcance de los niños. Podría causar
accidentes o daños.
El aparato contiene piezas pequeñas que pueden provocar un atragantamiento
si los niños las tragan accidentalmente.
Desconecte el adaptador de CA cuando no se use durante la medición.
El uso de accesorios no indicados en este manual puede afectar a la seguridad.
Si se produce un cortocircuito en la pila, puede calentarse y posiblementecausar quemaduras.
Permita que el aparato se ajuste al entorno circundante antes de usarlo(alrededor de una hora).
No se han realizado estudios clínicos en bebés recién nacidos o mujeres
embarazadas. No lo use en niños recién nacidos o mujeres embarazadas.
No toque las pilas, la clavija de CC y al paciente simultáneamente. Podría
ocasionar una descarga eléctrica.
No lo hinche sin colocar el brazalete en la parte superior del brazo.

Contraindicaciones
A continuación se indican precauciones para el uso correcto del aparato.
No aplique el brazalete a un brazo con otro aparato eléctrico médico acoplado.
Es posible que el aparato no funcione correctamente.
Las personas con un grave déficit circulatorio en el brazo deben consultar a un
médico antes de utilizar el dispositivo, para evitar problemas médicos.
No autodiagnostique los resultados de medición ni inicie el tratamiento usted
mismo. Consulte siempre a su médico para la evaluación de los resultados y el
tratamiento.
No aplique el brazalete a un brazo con una herida sin curar.
No aplique el brazalete a un brazo que está recibiendo un goteo intravenoso otransfusión sanguínea. Puede provocar lesiones o accidentes
Español3

No utilice el aparato en donde haya gases inflamables como, por ejemplo, gases
anestésicos. Puede provocar una explosión.
No utilice el aparato en entornos con alta concentración de oxígeno, tal como lascámaras de oxígeno de alta presión o tienda de oxígeno. Puede provocar un
incendio o explosión.

Español

4Partes de identificación de componentes
Pantalla

Tapón delconector de aire

Enchufeadaptadorde CAJack de CC

Zócalo de aire

Botón START
Manguerade aireMarca de posición
de la arteria

Rango de colocación

Marca índice
BrazaleteCompartimiento de pilas

Cubierta de pilas

Pilas de 1,5 V(R6P, LR6 o AA)

Indicadores de la pantalla

Promedio

MEMORIA
Símbolo LI/FibA
Presión
sistólica%LI/FibA
Símbolo de error de ajuste del brazalete

Presión
diastólica

Símbolo de error por movimient
Marca de corazón

Frecuenciadel pulso

Indicador de clasificación de OMS
e indicador de la barra de presión
Indicador de pilas
Español5

Símbolos
Símbolos impresos en la carcasa del aparato
Símbolos
SYS

Función / Significado
Aparato en “standby” y botón de encendido
Presión sanguínea sistólica indicada en mmHg.

DIA

Presión sanguínea diastólica indicada en mmHg.

PUL

Frecuencia del pulso por minuto
Guía para instalación de pilas
Corriente continua
Tipo: BF; El aparato, el brazalete y tubo están diseñados para
proporcionar protección especial contra electrochoques.
Etiqueta de aparatos médicos según directiva de CE
Representante UE
FabricanteFecha de fabricación

IP

Símbolo de protección internacional
Etiqueta de WEEE

SN

Número de serie
Consulte el manual/folleto de instrucciones
Polaridad del jack de CC
Manténgase seco

Símbolos que aparecen en la pantalla
SímbolosFunción / Significado
Acción recomendada
Aparece cuando la medición está en
curso. Este indicador parpadea cuando La medición está en curso.
se detecta el pulso.
Símbolo LI/FibA
Este indicador aparece cuando se
detectan latidos irregulares del corazón.
Puede iluminarse cuando se detecta una
vibración muy ligera, como un escalofrío
o un temblor.
La lectura puede ofrecer unvalor incorrecto. Vuelva a
Símbolo que aparece cuando se detecta
medir otra vez. Quédese
un movimiento del cuerpo o el brazo.
muy tranquilo durante la
medición.

Español

6

Símbolos
SímbolosFunción / Significado
Símbolo que aparece durante la
medición cuando el brazalete no estábien ajustado.

Acción recomendada
La lectura puede ofrecer unvalor incorrecto. Coloque elbrazalete correctamente, yvuelva a medir otra vez.

Índice detectado de LI/FibA en la memoria
Número detectado de
LI/FibA en la memoria
x 100 [%]
%LI/FibA =Número total
Las mediciones previas almacenadas enla MEMORIA.
Datos medios
PILAS CARGADAS
Indicador de la potencia de las pilas
durante la medición.
PILAS GASTADAS
Las pilas son débiles cuando parpadea
esta marca.
Presión sanguínea inestable debido al
movimiento durante la medición.

Reemplace todas las pilas
con unas nuevas, cuando
destelle esta marca.
Vuelva a medir otra vez.
Quédese muy tranquilo
durante la medición.

Los valores sistólico y diastólico seencuentran dentro de 10 mmHg.
El valor de presión no ha aumentado
Coloque el brazalete
durante la inflación.
correctamente, y vuelva a
El brazalete no está colocado
medir otra vez.
correctamente.
ERROR DE VISUALIZACIÓN DEL PULSOEl impulso no se ha detectado
correctamente.
Extraiga las pilas, presione elbotón START y a
Error interno del monitor de presiones
continuación vuelva a colocar
sanguíneas
las pilas. Si el error sigue
apareciendo, póngase en
contacto con su distribuidor.

Español7

Modo de operación
1. Medición normal con los datos almacenados
Presione el botón START . Se mide la presión sanguínea y sus datos se almacenan
en el aparato. El aparato puede almacenar los últimos 60 datos automáticamente enla memoria.

2. Revisión de los datos de la memoria En "standby",mantenga presionado
el START botón

Cuando no se visualice nada, mantengapresionado el botón START .
Suelte el botón cuando se muestren los datos
promedio.
El número de dato y los datos guardados semuestran automáticamente en el orden desdela última medición.
Para más detalles acerca de la revisión de datos,
consulte “Revisión de los datos de la memoria”.

3. Borrado de datos los datos almacenados en la memoria
Cuando apague el aparato, pulse y mantenga pulsado el".
botón _START_ hasta que se muestre "Seleccione "" para borrar los datos.
.
El dato se borra cuando parpadea la marca
El aparato se apaga automáticamente.

4. Medición con la presión sistólica deseada
Véase página 15 para la medición con la presión sistólica deseada.

Español

8

STARTUso del monitor
Instalación/Cambio de las pilas

Paso 1

1. Retire la cubierta de pilas.

Paso 3

2. Retire las pilas usadas e inserte lasnuevas en el compartimiento tal como
se describe, respetando laspolaridades (+ y -) de las pilas.
Utilice solo las pilas R6P, LR6 o AA.

Paso 1

Paso 2

3. Cierre la cubierta de pilas.

PRECAUCIÓN
Coloque las pilas en el compartimiento de las pilas, tal como se muestra en la figura.
De lo contrario, el aparato no funcionará.
Cuando la marca(marca de PILAS GASTADAS) parpadea en la pantalla, reemplace
todas las pilas por las nuevas. No mezcle pilas viejas con las nuevas, dado que esto
puede acortar la vida útil de las mismas o causar un mal malfuncionamiento delaparato.
Sustituya las pilas al menos dos segundos después de que se apague el aparato.
(marca de PILAS GASTADAS) aparece incluso después de sustituir las pilas,
Si la marca
realice una medición de la presión sanguínea. El aparato podría entonces reconocerlas pilas nuevas.
La marca(marca de PILAS GASTADAS) no aparece cuando las pilas están totalmenteagotadas.
La vida útil de las pilas varía con la temperatura ambiente y puede acortarse atemperaturas bajas. Por lo general, cuatro pilas LR6 nuevas duran
aproximadamente un año si se usan para realizar mediciones dos veces al día.
Utilice solamente las pilas especificadas. Las pilas suministradas con el aparato sonpara probar el funcionamiento del monitor y pueden tener una vida útil más corta.
Retire las pilas cuando no vaya a usar el aparato durante un período prolongado.
Las pilas pueden tener fugas y causar un malfuncionamiento.

Español9

Uso del monitor
Conexión de la manguera de aire Zócalo de aire
Inserte firmemente el tapón del conectorde aire en el zócalo de aire.

Tapón del conector de aire

Conexión del adaptador de CA
Inserte el enchufe del adaptador de
CA en el jack de CC. A continuación,
inserte el adaptador de CA en una
toma de corriente.

Jack de CC

Utilice el adaptador de CA especificado.
Enchufe adaptador de CA(Véase página 21.)
Cuando desconecte el adaptador de CA de la toma de corriente, agarre elarmazón del adaptador de CA y tire de él para extraerlo de la toma de corriente.
Cuando desconecte el enchufe del adaptador de CA del monitor de presiones
sanguíneas, agarre el enchufe del adaptador de CA y tire de él para extraerlo delmonitor.

Español

10Uso del monitor
Selección del brazalete correcto
La selección de un brazalete de tamaño correcto es importante para la lectura
correcta. Si el brazalete no es de tamaño correcto, la lectura puede indicar un
valor incorrecto de la presión sanguínea.
El tamaño del brazo está impreso en cada brazalete.
El índicey el gama adecuada de colocación del brazalete le indicarán, si elbrazalete está colocado correctamente. (Refiérase a "Símbolos impresos en el
brazalete" en la página siguiente.)
Si el índice
se encuentra fuera de la gama, póngase en contacto con el
distribuidor local para adquirir el brazalete de reemplazo.
El brazalete es un producto consumible. Si se deteriora, adquiera uno nuevo.
Tamaño del brazo
Tamaño del brazalete recomendado
Nº de catálogo
31 cm a 45 cm
Brazalete grande de adulto
CUF-F-LA
22 cm a 42 cm
Brazalete de amplio rango
CUF-I
22 cm a 32 cm
Brazalete de adulto
CUF-F-A
Tamaño del brazo: La circunferencia en el bíceps.
Nota:
El UA-767S-W no se ha diseñado para usar un brazaletepequeño.

Cómo colocar el brazalete

Evite que la manga quede demasiado
apretada al remangarse.

1. Coloque el brazalete alrededor del brazo,
unos 1-2 cm por encima de la parteinterior del codo, tal como se muestra en
la figura. Coloque el brazalete encontacto directo con la piel, dado que laropa puede causar un pulso débil yprovocar error de medición.

Brazalet
1-2 cmManguera de aire
Marca deposición de la arteria

2. Asimismo, la constricción del brazo,
causada al remangarse la manga
apretada, no permitirá lecturas
correctas.

Abroche bien elcierre de velcro.

se encuentre
3. Confirme que el índicedentro de la gama de buen estado.
ÍndiceEspañol11y la rango de colocación

Uso del monitor
Símbolos impresos en el brazaleteSímbolosFunción/Significado

Acción recomendada
Ajuste la marca ● en la arteriadel brazo o de forma alineada conel dedo anular en la parte interiordel brazo.

●

Marca de posición de la arteria▲
REFÍndiceNúmero de catálogo
Rango de colocación adecuado para el
brazalete de adulto.
Está impreso en el brazalete de adulto.
Por encima del rango impreso en el
brazalete de adulto y el brazalete deamplio rango.
Rango de colocación adecuado para el
brazalete de amplio rango.
Está impreso en el brazalete de amplio
rango.
Rango de colocación adecuado para el
brazalete grande de adulto.
Está impreso en el brazalete grande deadulto.
Por debajo del rango impreso en el
brazalete de adulto y el brazalete deamplio rango.
Por debajo del rango impreso en el
brazalete de adulto grande.

Utilice el brazalete grande deadulto en lugar del brazalete deadulto o brazalete de amplio rango.

Utilice el brazalete de adulto envez del brazalete grande de adulto.

Número de loteRango de colocación
Brazalete grandede adulto

A

L

Brazalete deamplio rango

S

W

L

Brazalete deadultoS

A

L

Español

12

Uso del monitor
Para efectuar mediciones correctas
Tenga en cuenta lo siguiente para una óptima precisión en la medición de latensión arterial:
Siéntese cómodamente en una silla. Apoye su brazo sobre la mesa. No cruce laspiernas. Mantenga los pies en el suelo y enderece la espalda.
Relájese durante unos 5 a 10 minutos.
Coloque el centro del brazalete al mismo nivel que su corazón
Permanezca quieto y en silencio durante la toma de la presión.
No efectúe la medición inmediatamente después de un ejercicio físico o baño.
Descanse unos 20 ó 30 minutos antes de la medición.
Trate de medir su tensión arterial a la misma hora todos los días.

Medición
Durante la medición, es normal que se sienta el brazalete muy apretado (no se
alarme por esto).

Después de la medición
Después de la medición, presione el botón _START_ para apagar el aparato.
Retire el brazalete y registre sus datos de medición.
Nota: El aparato cuenta con la función de desconexión automática de la alimentación,que desconecta la alimentación automáticamente un minuto después de la
medición.
Deje que transcurran por lo menos tres minutos entre mediciones de la misma
persona.

Español

13

Mediciones
Antes de la medición, lea “Notas para mediciones correctas” en la página siguiente.

Medición normal
1.

Coloque el brazalete en su brazo (brazo
izquierdo preferiblemente).
Siéntese tranquilamente durante la medición.

2.

Presione el botón _START_.
Aparecen brevemente todas las secciones dela pantalla. A continuación, parece cero (0)parpadeando brevemente. Luego la indicaciónde la pantalla cambia como se muestra en la
figura derecha, al comenzar la medición. El
brazalete comienza a inflarse. Es normal que
el brazalete apriete fuertemente. Se visualiza
el indicador de barra de presión durante la
inflación en el borde izquierdo de la pantalla.
Nota: Si desea detener el inflado en cualquier
momento, presione el botón _START_
nuevamente.

3.

4.

5.

A nivel del corazón

Presione
Aparecen brevemente
todas las secciones
de la pantalla.

Aparece ceroComienza la inflación

Al término de la inflación del brazalete, ladeflación comienza automáticamente y la(marca de corazón) parpadea
marca
indicando que la medición está en curso. Unavez detectado el pulso, esta marca parpadeará
coincidiendo con el latido del corazón.
Nota: Si no se obtiene una tensión apropiada, el
aparato empieza nuevamente el inflado
automáticamente.
Para evitar la reinflación, lea el apartado“Medición con la presión sistólica
deseada” en la página siguiente.
Al término de la medición, se visualizan laslecturas de las tensiones sistólica y diastólica
y frecuencia del pulso.
El brazalete expulsa el aire remanente y sedesinfla completamente.
Presione el botón _START_ para apagar elaparato.
Después de un minuto de inactividad, el
aparato se apaga automáticamente.

Presionamiento

Medición
progresiva
Presión sistólica
Presión diastólica
Clasificación según la OMS

Frecuencia del pulso

Nota: Deje que transcurran por lo menos tres minutos
entre mediciones de la misma persona.
Español

14

START

Evacuación automática
del aire remanente.

Mediciones
Medición con la presión sistólica deseada
El UA-767S-W está diseñado para detectar el pulso, inflando automáticamente
el brazalete hasta el nivel de presión sistólica.
Utilice este método cuando la represurización se produzca de forma repetida, ocuando los resultados no se muestren incluso aunque la presión disminuya a
20 mmHg o menos.
1.

Coloque el brazalete en el brazo en el nivel
del corazón (preferiblemente el brazo
izquierdo).

2.

Pulse el botón _START_ .

3.

Mientras el cero parpadea, pulse ymantenga pulsado el botón _START_
hasta que aparezca un número
alrededor de 30 a 40 mmHg superior a lapresión sistólica esperada.

4.

Suelte el botón _START_ al alcanzar elvalor deseado, ya que la medición
comenzará. Luego continúe la medición desu presión sanguínea como se describe enla página anterior.

Al nivel delcorazón
Pulse el botón START
Suelte el botón cuando
se visualice la presión
sistólica deseada.

Mantenga presionado STARTel botónPara la medición, refiérase
a la página anterior.

Notas para mediciones correctas

Siéntese cómodamente. Coloque el brazo sobre una mesa con la palma hacia
arriba y el brazalete al mismo nivel que el corazón.
Relájese por unos cinco o diez minutos antes de iniciar la medición. Si seencuentra agitado o deprimido debido a un estrés emocional, la medición reflejará
este estrés con una lectura mayor (o menor) que la de su tensión arterial normal yla lectura del pulso generalmente será más rápida que la normal.
La tensión arterial de una persona varía constantemente, dependiendo de lo queestá haciendo y lo que ha comido. Lo que tome podrá tener un efecto muy fuertey rápido sobre su tensión arterial.
Las mediciones de este aparato se basan en el latido del corazón. Si usted tiene
un latido muy débil o irregular, este aparato puede tener dificultad en determinarsu tensión arterial.
En caso de que el aparato detecte una condición anormal, el mismo detendrá lamedición e indicará un símbolo de error en la pantalla. Vea la página 7 para ladescripción de los símbolos.
Español

15

Este aparato está diseñado para su uso en adultos. Consulte con su médico
antes de usar este aparato en un niño. Un niño no debe usar este aparato sin la
atención de un adulto.
El rendimiento del tensiómetro automático puede verse afectado por temperatura
y humedad excesivas o por la altitud.

Revisión de los datos de la memoria
Nota:
1.

Este aparato almacena en la memoria las últimas 60 mediciones.
Mantengapresionado START
Si no se muestra nada, pulse yel botónmantenga pulsado el botón _START_
Presión sistólica media
para recuperar los datos guardados.
Presión diastólica media

2.

Suelte el botón cuando se muestren losdatos promedio.

3.

El número de dato y los datos
guardados se muestranautomáticamente en el orden desde laúltima medición.

4.

La pantalla se apagará
automáticamente después de habermostrado todos los datos.

Frecuencia media del pulso
Promedio detodos los datos

Sistólica
Diastólica
Frecuenciadel pulso
Datos más recientes

Nota: Si pulsa el botón _START_ cuando
recupera los datos, el aparato se
apagará.

Sistólica
Diastólica
Frecuenciadel pulsoÚltimo dato (más antiguo)
Español16¿ Qué es el indicador de LI/FibA?
Si el monitor detecta un ritmo irregular durante las mediciones, aparecerá elindicador LI/FibA en la pantalla con los valores de medición.
Nota: Le recomendamos que consulte a su médico si ve el símbolo
LI/FibA con frecuencia.

¿A qué hace referencia FibA?
El corazón se contrae debido a corrientes eléctricas que surgen de él y envía
sangre por todo el cuerpo. La fibrilación auricular (FibA) se produce cuando las
corrientes eléctricas del atrio se confunden y ocasionan alteraciones en losintervalos del pulso. La FibA puede ocasionar estancamientos de sangre en elcorazón, lo que podría producir fácilmente coágulos de sangre, una de lascausas de las apoplejías y los infartos de corazón.

％LI/FibA%LI/FibA aparece como índice del LI/FibA detectado.
LI/FibA puede detectar no solamente interferencias tales como el movimiento
físico, sino también los latidos irregulares del corazón. Por tanto, lerecomendamos consultar con su médico si el nivel de %LI/FibA es alto.
Número detectado de
LI/FibA en la memoria
x 100 [%]
%LI/FibA =Número total
Visualización de %LI/FibA: %LI/FibA se visualiza al mostrar los valores medios.
(Refiérase a “2. Revisión de los datos de lamemoria” en “Modo de operación”.)%LI/FibA no se visualiza si el número almacenado
en la memoria es igual o inferior a seis.
Visualización de valores medios

Nivel 0%LI/FibA = 0

Nivel 1
%LI/FibA = 1 - 9

Nivel 2
Nivel 3
%LI/FibA = 10 - 24 %LI/FibA = 25 - 100

No se visualiza

Español

17

Indicador de barra de presión
El indicador monitoriza el progreso de la presión durante la medición.
Presurizando

Liberando aire

Inflación en progreso

Medición en progreso
Se ha completado la inflación

Indicador de clasificación de la OMS
Cada uno de los seis segmentos del indicador debarra corresponde a la clasificación de presión
sanguínea de la OMS que se describe en lasiguiente página.
Indicador de clasificación de la OMS
Hipertensión severa
Hipertensión moderada
Hipertensión leve
Normal alta
NormalOptimaEjemplo:

Hipertensión moderada

Hipertensión leve

: El indicador visualiza un segmento basado enel dato actual, correspondiente a laclasificación de la OMS.

Normal alta

Aspectos sobre la tensión arterial¿Qué es la tensión arterial?
La tensión arterial es la fuerza que ejerce la sangre contra las paredes de lasarterias. La tensión sistólica ocurre cuando el corazón se contrae, y la tensión
diastólica ocurre cuando el corazón se expande.
La presión sanguínea se mide en milímetros de mercurio (mmHg). La presión
arterial de una persona está representada por la tensión fundamental, que esla que se mide primero en la mañana cuando la persona está aún descansada
y antes de tomar alimentos.
Español18¿Qué es la hipertensión y cómo controlarla?
La hipertensión es la tensión arterial anormalmente alta, si no se corrige, puedecausar muchos problemas de salud, incluyendo ataques al corazón y apoplejía.
La hipertensión puede controlarse cambiando el estilo de vida, evitando elestrés y con medicamentos bajo la supervisión de un médico.
Para evitar la hipertensión o para mantenerla bajo control:
No fume.
Haga ejercicios con regularidad.
Reduzca la ingestión de sal y grasa.
Sométase a exámenes físicosMantenga un peso apropiado.
con regularidad.

¿Por qué debe medirse la tensión arterial en la casa?
La medición de la tensión arterial en una clínica o un consultorio médico puedecausar temor y, por lo tanto producir una lectura elevada, de 25 a 30 mmHg
mayor que la medida en la casa. La medición en casa reduce los efectos deinfluencias externas sobre las lecturas de tensión arterial, complementa laslecturas del médico y proporciona un historial más completo y exacto de latensión arterial.

Clasificación de tensión arterial según la OMS
OMS ha establecido las normas
para la evaluación de la tensión
arterial, tal como se muestra en el
gráfico.

Variaciones de tensión arterial
La tensión arterial de una persona
varía considerablemente en basediaria y estacional. La misma puedevariar en 30 a 50 mmHg
dependiendo de diversas condiciones durante el día. En personas hipertensas,
estas variaciones son aun más notables.
La tensión arterial normalmente se eleva cuando se está trabajando o jugando ybaja a sus niveles más bajos al dormir. Por lo tanto, no se preocupe demasiado
de los resultados de una sola medición.
Mídase la tensión arterial a la misma hora
todos los días usando el procedimiento
descrito en este manual y tenga
conocimiento de su tensión normal.
Lecturas múltiples proporcionarán unhistorial de tensión arterial más completo.
Asegúrese de anotar la fecha y hora al
registrar su tensión arterial. Consulte consu médico para interpretar los datos de su
tensión arterial.

Español

19

Localización y corrección de fallos
Problema

Causa probableNo aparece nada Las pilas se han agotado.
en la pantalla,
aun cuando se
Los terminales de las pilas noconecta laestán en posición correcta.
alimentación.
El brazalete no
se infla.

El aparato no
mide la presión
arterial. Las
lecturas son
demasiado altas
o demasiado
bajas.

Otros

Acción recomendada
Reemplace todas las pilas
por otras nuevas.
Coloque las pilas con sus
terminales negativo y positivo
coincidiendo con los indicadosen la banda inflable.

El voltaje de las pilas está bajo.
Parpadea la marca(marca deReemplace todas las pilas
PILAS GASTADAS). Si las pilas
por otra nuevas.
están agotadas completamente,
no aparece esta marca.
El brazalete no está ajustado
Ajuste el brazaletedebidamente.
correctamente.
Asegúrese de mantenerse
Movió su brazo o cuerpo durante
muy quieto y en silencio
la medición.
durante la medición.
Siéntese cómodamente ymanténgase quieto. Coloque
El brazalete no está posicionado el brazo sobre una mesa conla palma hacia arriba y elcorrectamente.
brazalete al mismo nivel que
el corazón.
Si el latido de su corazón es
muy débil o irregular, elaparato tendrá dificultad en
determinar su tensión arterial.
El valor se diferencia al medido Consulte el capítulo 3, ¿Por
qué debe medirse la tensión
en una clínica o un consultorio
arterial en la casa?
médico.
Retire las pilas. Colóquelas
nuevamente de forma
correcta e intente la medición
nuevamente.

Nota: Si las acciones descritas anteriormente no solucionan el problema, póngase
en contacto con su distribuidor. No abra el aparato ni intente repararlo, ya que
cualquier intento de reparación invalidará la garantía.

Español

20

Mantenimiento
No abra el aparato, ya que éste contiene componentes eléctricos delicados y
una unidad de aire compleja que podrían dañarse. Si no puede solucionar el
problema aplicando las instrucciones de localización y corrección de averías,
solicite el servicio de reparación a su proveedor o al grupo de servicio de A&D.
El grupo de servicio de A&D proveerá información técnica, repuestos y unidadesa los proveedores autorizados.
Este aparato está diseñado y fabricado para una vida de servicio prolongada.
No obstante se recomienda someter el monitor a la inspección cada 2 años afin de asegurar el funcionamiento y la precisión adecuadas. Sírvase ponerse
en contacto con su distribuidor autorizado o A&D para su mantenimiento.

Datos técnicos
Tipo
Método de medición
Rango de medición

Precisión de medición
Alimentación
Número de mediciones

Clasificación

Prueba clínica

EMD
Almacenamiento de datos
Condiciones ambientales
de funcionamiento
Condiciones de transporte
almacenamiento

UA-767S-W
Medición oscilométrica
Tensión: 0 a 299 mmHg
Tensión sistólica: 60 a 279 mmHg
Tensión diastólica: 40 a 200 mmHg
Pulso:
40 a180 latidos/minuto
Tensión: ±3 mmHg
Pulso:±5%
4 pilas de 1,5V (R6P, LR6 o AA) o
un adaptador de CA (TB-233C) (No incluido)
Aprox. 700 veces LR6 (pilas alcalinas)
Aprox. 200 veces R6P (pilas de manganeso)
Con un valor de presión de 180 mmHg,
temperatura ambiente 23 ºC.
Equipo ME alimentado internamente (suministrado
por pilas) /Class II (suministrado por adaptador)
Modo de operación continuo
De acuerdo con ISO81060-2: 2013
En el estudio de validación clínica, se determinó lapresión arterial diastólica en 85 sujetos mediante
K5.
IEC 60601-1-2: 2014
Las últimas 60 medidas
Temperatura de +10 a +40°C / RH de 15 a 85% /de 800 a 1060 hPa
Temperatura de -20 a +60°C / RH de 10 a 95 % /Presión de 700 a 1060 hPa
Español21

Dimensiones
Peso
Protección de admisión
Pieza aplicada
Vida útil

Aprox. 140 [An] x 60 [Al] x 105 [Pr] mm
Aprox. 245 g, excluyendozlas pilas
Dispositivo: IP20
Brazalete Tip BF
Aparato: 5 años (cuando se usa seis veces al día)
Brazalete: 2 años (cuando se usa seis veces al día)
Adaptador de CA: 5 años (cuando se usa seis veces al día)

Accesorio: Adaptador de CA, El adaptador se conecta a la fuente de alimentación
modelo TB-233C
casera. Sírvase tener contacto con la agencia A&D
local para comprarlo.
Es necesario inspeccionar o sustituir el adaptador CA
periódicamente.
Símbolos impresos en el adaptador de CA
SímbolosFunción / Significado
Solo para uso en interiores
Aparato Clase II
Fusible térmico
FusibleEtiqueta de la directiva CE
Etiqueta de homologación EAC
Polaridad del enchufe del adaptador de CA
Accesorios vendidos por separado
BrazaleteNúmero de catálogo
Tamaño del brazaleteTamaño del brazo
CUF-F-LA
Brazalete grande de adulto
31 cm a 45 cm
CUF-I
Brazalete de amplio rango
22 cm a 42 cm
CUF-F-A
Brazalete de adulto
22 cm a 32 cm
Tamaño del brazo: circunferencia en el bíceps.
Adaptador de CA

Nota:

Número de catálogo
TB-233C

Clavija
Tipo CLas especificaciones están sujetas a cambio sin previo aviso.
La clasificación IP hace referencia a los grados de protección otorgados por
los materiales contenedores según la norma CEI 60529. Este dispositivo está
protegido contra objetos extraños sólidos de 12 mm de diámetro y más, por
ejemplo, un dedo. Este dispositivo no está protegido contra el agua.

Español22

Indice
Introduzione .......................................................................... 2
Norme Precauzionali .............................................................. 2
Precauzioni ............................................................................ 2
Identificazione delle parti ...................................................... 5
Simboli .................................................................................. 6
Modalità operative ................................................................. 8
Preparazione del misuratore .................................................. 9
Installazione / Sostituzione delle batterie .......................... 9
Collegamento del bracciale allo strumento......................... 10
Collegamento dell’alimentatore a rete elettrica ............... 10
Come scegliere il bracciale corretto................................. 11
Applicazione del bracciale ............................................... 11
Consigli per l’esecuzione di misurazioni corrette ............ 13
Durante la misura ........................................................... 13
Dopo la misura ............................................................... 13
Misurazione della pressione ................................................. 14
Misurazione standard ..................................................... 14
Misurazione con impostazione della pressione di gonfiaggio desiderata.... 15
Note per una misurazione corretta .................................. 15
Richiamo delle misure memorizzate.......................................... 16
Cosa è l’indicatore IHB/Afib? .................................................... 17
Cosa è l’AFib? ...................................................................... 17％IHB/AFib ........................................................................... 17
Indicatore dinamico della pressione .......................................... 18
Indicatore di classificazione dell’O.M.S. ..................................... 18
Parlando di Pressione Arteriosa ............................................ 18
Cosa si intende per Pressione Arteriosa? ......................... 18
Cosa si intende per Ipertensione Arteriosa? ..................... 19
E come si controlla? ........................................................ 19
Perché è importante l’automisurazione al proprio domicilio? .... 19
Classificazione della pressione secondo le linee guida “O.M.S.” ..... 19
Localizzazione guasti .......................................................... 20
Manutenzione ...................................................................... 21
Dati tecnici .......................................................................... 21

Italiano1Introduzione
Gentile Cliente, la ringraziamo per aver acquistato il Misuratore Elettronicodella Pressione Arteriosa A&D. Concepito per una facilità di uso ed alta
precisione, il nostro prodotto la aiuterà a controllare la pressione arteriosa
quotidianamente.
Raccomandiamo di leggere con cura questo manuale prima di utilizzare il
misuratore per la prima volta.

Norme Precauzionali
Il dispositivo è conforme alla Direttiva Europea CEE 93/42 per i Prodotti adutilizzo medico. Quanto sopra è evidenziato dal marchio di conformità.
(0123: Numero di riferimento per l'ente che riceve la notifica)
Il presente dispositivo deve essere utilizzato da parte di persone adulte, non dineonati o bambini.
Ambiente di utilizzo. Il dispositivo è idoneo per l’utilizzo da parte dell’utente inambienti di assistenza sanitaria domestici.
Il dispositivo è progettato per misurare la pressione sanguigna e la frequenza
cardiaca di soggetti umani a fini diagnostici.

Precauzioni
Il presente dispositivo è stato realizzato utilizzando componenti di precisione. Si
raccomanda di evitare condizioni estreme di temperatura, umidità, luce solare
diretta, urti o polvere.
Pulire il dispositivo e il bracciale utilizzando un panno morbido e asciutto o inumidito
con acqua e detergente neutro. Evitare l’utilizzo di alcool, benzene, solventi o altre
sostanze chimiche aggressive per pulire il dispositivo o il bracciale.
Evitare di piegare eccessivamente il bracciale o di riporre il tubo avvolto
strettamente per lunghi periodi, poiché quanto sopra potrebbe ridurre la durata deicomponenti.
Fare attenzione a evitare lo strangolamento accidentale di bambini o infanti con iltubo e il cavo.
Durante la misurazione non attorcigliare il tubo prolunga, in quanto potrebbero
verificarsi delle lesioni dovute alla pressione continua nel bracciale.
Il dispositivo e il bracciale non sono impermeabili. Evitare il contatto con pioggia,
sudore e acqua.
Nel caso in cui il dispositivo sia utilizzato in prossimità di apparecchi televisivi, fornia microonde, telefoni cellulari, macchinari a raggi X o di altri dispositivi con forti
campi elettrici, le misurazioni potrebbero non risultare precise.
I dispositivi per la comunicazione wireless, come i dispositivi delle reti domestiche, itelefoni cellulari, i telefoni cordless e le loro stazioni base, i walkie-talkie possono
interferire con il funzionamento di questo Monitor della Pressione Sanguigna. Di
conseguenza, è consigliabile mantenere una distanza minima di 30 cm da talidispositivi.
Italiano 2

Al riutilizzo del dispositivo, verificare che sia pulito.
Le apparecchiature, le parti e le batterie usate non devono essere trattate quali
normali rifiuti domestici. Devono quindi essere smaltite in conformità alle normativein vigore.
Se si utilizza l'alimentatore, assicurarsi che sia possibile scollegarlo dalla presa
elettrica in caso di necessità.
Non modificare il dispositivo. Questa operazione può causare incidenti o danni al
dispositivo stesso.
Per misurare la pressione sanguigna è necessario avvolgere strettamente il
bracciale intorno al braccio in modo da interrompere temporaneamente il flusso
sanguigno nell'arteria. Ciò può causare dolore, intorpidimento o lasciare
temporaneamente un segno rosso sul braccio, in particolare se la misurazione
viene ripetuta più volte di seguito. Il dolore, l'intorpidimento o i segni rossi che
possono eventualmente verificarsi scompaiono dopo breve tempo.
Misurazioni della pressione sanguigna troppo frequenti possono provocare danni alpaziente a causa delle interferenze con il flusso sanguigno. In caso di uso ripetutodel dispositivo, assicurarsi che il funzionamento del misuratore non comporti una
prolungata compromissione della circolazione sanguigna.
Se si è stati sottoposti a mastectomia, rivolgersi a un medico prima di utilizzare il
dispositivo.
Evitare che i bambini usino il dispositivo senza sorveglianza e tenere il dispositivo
fuori dalla loro portata. In caso contrario, si potrebbero causare incidente o danni.
Presenta piccole parti che possono causare soffocamento se ingerite
inavvertitamente dai bambini.
Scollegare l’adattatore C.a. quando non lo si usa durante la misurazione.
L’utilizzo di accessori non indicati in questo manuale potrebbe compromettere la
sicurezza.
In caso di cortocircuito della batteria, la suddette potrebbe riscaldarsi e causare
potenzialmente delle ustioni.
Consentire al dispositivo di adattarsi all’ambiente circostante, prima dell’uso (circa
un’ora).
Non sono stati condotti test clinici sui neonati e le donne in gravidanza. Non usare
su neonati o donne in gravidanza.
Non toccare contemporaneamente le batterie, la presa jack C.c. e il paziente. Si
potrebbero causare scosse elettriche.
Non gonfiare senza avvolgere il bracciale attorno al braccio.

Controindicazioni
Di seguito sono indicate le precauzioni da osservare per il corretto utilizzo del dispositivo.
Non applicare il bracciale a un paziente già collegato a un altro dispositivo
elettromedicale. Il dispositivo potrebbe non funzionare correttamente.
Le persone affette da grave deficit circolatorio al braccio devono consultare un
medico prima di utilizzare il dispositivo, al fine di evitare problemi medici.
Non eseguire auto-diagnosi dei risultati delle misurazioni, né iniziare trattamenti inautonomia. Per la valutazione dei risultati ed eventuali trattamenti, rivolgersi
sempre al medico.
Non applicare il bracciale se il braccio presenta ferite non cicatrizzate.
Italiano 3

Non applicare il bracciale in caso di terapia con flebo o trasfusioni di sangue
eseguite al braccio del paziente. Pericolo di lesioni o incidenti.
Non utilizzare il dispositivo in ambienti in cui siano presenti gas infiammabili, quali i
gas anestetici. Pericolo di esplosione.
Non utilizzare il dispositivo in ambienti con concentrazioni molto elevate di ossigeno,
quali camere iperbariche o tende a ossigeno. Pericolo di incendio o esplosione.

Italiano 4

Identificazione delle parti
Display a cristalli liquidi

Spinadell’alimentatore

Spina innesto aria

Jack a CC
Pulsante START(AVVIO)

Presa d’aria

Indicatore di scelta delbracciale adeguato

Tubo prolunga
Simbolo di indicazione
posizione arteria brachiale

Indicatore di posizione
Bracciale
Vano batterie

Coperchio

Batterie da 1,5 V(R6P, LR6 o AA)

Particolari del Display

Valori medi

MEMORIA
Simbolo IHB/AFib
Pressione
sistolica%IHB/AFib
Simbolo di errore diposizionamento del bracciale

Pressione
diastolica

Simbolo di errore di movimento
Simbolo del cuore(Indicazione misura in corso)

Frequenza
cardiaca

Indicatore di classificazione OMS e
indicatore di pressione
Indicatore batterie
Italiano 5

Simboli
Simboli impressi sulla confezione del dispositivo
Simboli
SYS
DIA
PUL

EC

REP

IP

SN

Funzione/Significato
Posizionare l’apparecchio in modalità standby, quindi collegarlo
Pressione arteriosa sistolica in mmHg
Pressione arteriosa diastolica in mmHg
Battiti cardiaci al minuto
Guida all'installazione delle batterie
Corrente continua
Tipo BF: dispositivo, bracciale e tubi sono stati progettati al fine digarantire una protezione particolare contro eventuali scosse elettriche.
Etichetta Direttiva CE sui dispositivi medici
Rappresentante UE
Produttore
Data di produzione
Simbolo internazionale di protezione
Etichetta Direttiva RAEE (direttiva Europea di smaltimento dei
dispositivi elettrici ed elettronici)
Numero di matricola
Consultare il manuale/l'opuscolo di istruzioni
Polarità del jack a CC
Tenere asciutto

Simboli visualizzati sul display
SimboliFunzione/Significato
Azione raccomandata
Indicazione visualizzata durante la fase La misurazione è in corso.
di misurazione. Lampeggia durante il
Rimanere quanto più
rilevamento della frequenza cardiaca.
possibile fermi e rilassati.
Simbolo IHB/AFib
Indicazione visualizzata se viene rilevato
un battito cardiaco irregolare. Potrebbe
illuminarsi se viene rilevata una
leggerissima vibrazione come un brivido
o un tremore.
La lettura potrebbe
presentare un valore
Simbolo visualizzato se viene rilevato un incorretto.
Ripetere la misurazione.
movimento corporeo o del braccio.
Rimanere quanto possibile
fermi e rilassati.
Italiano 6

Simboli
SimboliFunzione/Significato
Simbolo visualizzato in fase dimisurazione se il bracciale è allentato.

Azione raccomandata
La lettura potrebbe presentare
un valore incorretto.
Applicare e stringere
adeguatamente il bracciale eripetere la misurazione.

Frequenza rilevata di IHB/AFib in memoria
Numero di IHB/AFibrilevati in memoria
x 100 [%]
%IHB/AFib =Numero totale
Misurazioni precedenti MEMORIZZATE
Valori medi
BATTERIE CARICHE
Indicatore di stato di carica delle batterie
durante la misurazione
Qualora l’indicatore lampeggi,
BATTERIE SCARICHE
sostituire tutte le batterie.
Indicatore di batterie in esaurimento
Ripetere la misurazione.
Valori della pressione non corretti acausa di movimenti durante la
Rimanere quanto possibile
misurazione
fermi e rilassati.
Differenza tra pressione sistolica ediastolica inferiore a 10 mmHg
Pressione di gonfiaggio non sufficiente
Applicare e stringere
Il bracciale non è applicato in modo
adeguatamente il braccialecorretto
e ripetere la misurazione.
ERRORE VISUAL. POLSO
Il battito cardiaco non è stato rilevato inmodo corretto
Rimuovere le batterie epremere il pulsante START
Errore interno del misuratore di(AVVIO), quindi reinstallare lepressione arteriosa
batterie. Se l’errore persiste,contattare il rivenditore.

Italiano 7

Modalità operative
1. Misurazione normale con memorizzazione dei dati
Premere il pulsante START . La pressione arteriosa viene misurata e i dati sono
memorizzati nel dispositivo. Sono memorizzati automaticamente gli ultimi 60 valoridi pressione.

2. Richiamo dei dati
Quando non è visualizzato niente, tenere
premuto il pulsante _START_.

In standby,
tenere premuto il pulsante
START

Rilasciare il pulsante quando si visualizza
la media dei dati.
Vengono visualizzati automaticamente il
numero dei dati e il numero dei dati
memorizzati a partire dall'ultima
misurazione.
Per i dettagli su come richiamare i dati,
consultare “Richiamo delle misurememorizzate”.

3. Per eliminare tutti i dati memorizzati
Durante lo spegnimento del dispositivo, premere e tenere
premuto il pulsante _START_ (AVVIO) sino a visualizzare la
scritta "".
" per cancellare i dati.
Selezionare "lampeggia.
I dati sono cancellati quando il simbolo
Il dispositivo si spegne automaticamente.

START

4. Misurazione con impostazione della pressione di gonfiaggio
desiderata
Vedere descrizione completa a pagina 15.
Italiano 8

Preparazione del misuratore
Installazione / Sostituzione delle batterie
1. Rimuovere il coperchio del vano batterie.
2. Rimuovere le batterie esaurite ed
inserire le nuove batterie, rispettando
le polarità (+ e -) come indicato nella
figura a lato.
Utilizzare esclusivamente batterie R6P,
LR6 o AA.

Procedura 1
Procedura 3

Procedura 1

Procedura 2

3. Chiudere il coperchio del vano batterie.

ATTENZIONE
Inserire le batterie come indicato nel vano porta batterie. In caso contrario,
l'apparecchio non funzionerà.
Quando il simbolo(BATTERIE SCARICHE) lampeggia sul display, sostituire tutte
le batterie. Non mischiare batterie nuove con batterie usate, per evitare di ridurre
la durata delle batterie o causare malfunzionamenti del dispositivo.
Spento il dispositivo, attendere almeno due secondi per sostituire le batterie.
(BATTERIE SCARICHE) viene visualizzato anche dopo averSe il simbolo
sostituito le batterie, eseguire una misurazione della pressione arteriosa. Il
dispositivo è in seguito in grado di riconoscere le batterie nuove.
Il simbolo(BATTERIE SCARICHE) non viene visualizzato in caso di batteriecompletamente esaurite.
La durata delle batterie varia in base alla temperatura ambiente e potrebbe risultare
ridotta in caso di esposizione a basse temperature. In genere, quattro batterie LR6nuove durano circa un anno, quando vengono utilizzate due volte ogni giorno per lamisurazione.
Utilizzare solamente le batterie del tipo specificato. Le batterie fornite in dotazione al
dispositivo sono a solo scopo di prova delle prestazioni del monitor e la loro durata
potrebbe essere breve.
Togliere le batterie nel caso in cui il dispositivo non sia utilizzato per lungo tempo.
Le batterie potrebbero infatti essere soggette a perdite, causando quindi dei
malfunzionamenti.

Italiano 9

Preparazione del misuratore
Collegamento del bracciale allo strumento
Inserire fermamente la spina innesto
aria nella presa d’aria.

Presa d’aria

Spina innesto aria

Collegamento dell’alimentatore a rete elettrica
Inserire la spina dell’alimentatore nel jacka CC.
Inserire successivamente l’alimentatore
nella presa elettrica.
Jack a CC
Utilizzare l’alimentatore a rete elettrica
Spina dell’alimentatore
specificato. ( Consultare pagina 21.)Per scollegare l’alimentatore dalla presa elettrica, afferrare e rimuovere il
corpo dell’alimentatore dalla presa.
Per scollegare la spina dell’alimentatore dal Misuratore Elettronico Della
Pressione Arteriosa, afferrare ed estrarre la spina dell’alimentatore dal
misuratore.

Italiano10Preparazione del misuratore
Come scegliere il bracciale corretto
L’uso della dimensione corretta di un bracciale è importante per una lettura precisa.
Se il bracciale non presenta la dimensione corretta, la lettura potrebbe produrre unvalore incorretto della pressione sanguigna.
La dimensione del braccio è stampata su ogni bracciale.
Il simbolo
e l'indicatore di scelta del bracciale adeguato determinano se è stato
utilizzato il bracciale corretto. (Consultare "Simboli stampati sul bracciale" alla pagina
seguente.)
Se il simbolo
è posizionato fuori dal range, si raccomanda l’acquisto di un
bracciale appropriato (consultare la tabella sottostante).
Il bracciale è un prodotto di normale consumo. In caso di usura, acquistarne uno
nuovo.
Dimensione raccomandata delNumero di catalogo
bracciale
31 cm a 45 cm
Bracciale grande per adulti
CUF-F-LA
22 cm a 42 cm
Bracciale ampio
CUF-I
22 cm a 32 cm
Bracciale per adulti
CUF-F-A
Misura del braccio: corrisponde alla circonferenza del bicipite.
Nota: il UA-767S-W non è indicato per l'uso con il bracciale piccolo.
Dimensione del braccio

Applicazione del bracciale

Arrotolare la manica evitando di
stringere troppe il braccio

1. Avvolgere il bracciale intorno al braccio
superiore, a circa 1 - 2 cm sopra la piegadel gomito, come indicato nella figura alato. Applicare il bracciale direttamente a
contatto con la pelle, in quanto i vestiti
possono causare errori nella misura.

Bracciale
1-2 cmTubo prolunga
Simbolo di indicazione
posizione arteria brachiale

2. La compressione al braccio causata da
un'avvolgimento troppo stretto dellamanica della camicia, puo` causare una
misura errata.

Avvolgere il bracciale
saldamente

3. Verificare che il simboloè
all'interno del range di selezione.
Simbolo

Italiano

11

e gamma adatta adeguataPreparazione del misuratore
Simboli stampati sul bracciale
SimboliFunzione/Significato●

Simbolo di indicazione posizione arteria
brachiale▲
REF

Azione raccomandata
Posizionare il simbolo ●sull’arteria del braccio o inallineamento con il ditoanulare nella parte interna delbraccio.

IndiceNumero di catalogo
Gamma adatta a bracciale per adulti
È stampato sul bracciale per adulti
Utilizzare il bracciale grande
per adulti anziché il bracciale
per adulti o il bracciale ampio.

Gamma superiore stampata sul bracciale
per adulti e sul bracciale ampio.
Gamma adatta a bracciale ampio.
È stampato sul bracciale ampio.
Gamma adatta a bracciale grande per
adulti.
È stampato sul bracciale grande per adulti
Gamma inferiore stampata sul bracciale
per adulti e sul bracciale ampio.

Utilizzare il bracciale per adulti
anziché il bracciale grande per
adulti.

Gamma inferiore stampata sul bracciale
grande per adulti.
Numero di lotto

Indicatore di scelta del bracciale adeguato
Bracciale grande
per adulti

AL

Bracciale ampio

S

W

L

Bracciale per adulti

S

A

L

Italiano12Preparazione del misuratore
Consigli per l’esecuzione di misurazioni correttePer garantire la migliore precisione delle rilevazioni suggeriamo di:
Sedersi comodamente su una sedia. Appoggiare il braccio sul tavolo. Non incrociare
le gambe. Tenere i piedi appoggiati al suolo e raddrizzare il dorso.
Prima di procedere alla misurazione rilassarsi per 5 -10 minuti.
Avvolgere il bracciale allo stesso livello del cuore.
Durante la misurazione, mantenersi fermi e tranquilli.
Non effettuare la misurazione subito dopo aver eseguito esercizi fisico o aver fatto
il bagno. In tali casi, riposare per venti-trenta minuti prima di procedere alla
misurazione.
Si consiglia di misurare la pressione arteriosa tutti i giorni alla stessa ora.

Durante la misura
Durante la misurazione, è normale sentire il bracciale a pressione stringere
fortemente (Non preoccuparsi).

Dopo la misura
Dopo la misurazione, premere il pulsante di START per spegnere il dispositivo.
Rimuovere il bracciale e registrare i vostri dati.
Nota: L'apparecchio prevede una funzione d'interruzione automatica dell'alimentazione,che disattiva automaticamente l'alimentazione, un minuto dopo la misurazione.
E’ consigliabile lasciar passare almeno 3 minuti prima di effettuare una nuova
misura sulla stessa persona.

Italiano13

Misurazione della pressione
Prima di eseguire la misurazione, consultare “Note per una misurazione
corretta“ alla pagina seguente.

Misurazione standard
1.

Applicare il bracciale preferibilmente sul braccio sinistro
Al livello delcuoreallo stesso livello del cuore. Durante la misurazione
rimanere quanto più possibile fermi e rilassati.
2. Premere il pulsante START .
Premere STARTPer un attimo verranno visualizzati tutti i segmenti
sul display e in un secondo tempo, comeTutte le parti delmostrato nella figura a lato, verrà visualizzato lo 0
display vengono(zero) lampeggiante per alcuni istanti.
visualizzate
Successivamente il bracciale inizierà a gonfiarsi(è normale sentire stringere decisamente) e unindicatore dinamico a barre sarà visualizzato
Visualizzazione
nell'area a sinistra del display, mostrando
zerol’incremento della pressione di gonfiaggio.
Inizio della fase diNota: Nel caso in cui, in qualsiasi momento, si
gonfiamento
desideri sospendere la misurazione,
premere nuovamente il pulsante START .
3. Una volta completata la fase di pressurizzazione
lo sgonfiamento sarà automatico ed il simbolo
Pressurizzazione(simbolo del cuore) inizierà a lampeggiare,
indicando che la misurazione è in corso. Una
volta rilevate le pulsazioni, il simbolo del cuore
lampeggerà in sincronia con il battito cardiaco.
Nota: Nel caso in cui non si ottenga una pressione digonfiaggio adeguata, il misuratore inizieràautomaticamente a rigonfiare il bracciale.
Per evitare il rigonfiaggio, consultare“Misurazione con impostazione della pressione
di gonfiaggio desiderata” alla pagina successiva.
Misurazione
4. Una volta completata la misurazione, saranno
in corso
visualizzati i valori relativi alla pressione sistolica,
diastolica e alla frequenza cardiaca, insieme a unsegmento lampeggiante in corrispondenza della
Pressione sistolica
scala dei valori dell’O.M.S. Il bracciale espellerà
Pressione diastolica
l’aria residua e si sgonfier completamente.
Classificazione OMS
5. Premere il pulsante START per spegnere il
Frequenza cardiaca
dispositivo.
Dopo un minuto di inattività, il dispositivo si
spegnerà automaticamente.
Espulsione automatica
dell’aria
residua nel bracciale
Nota: E’ consigliabile lasciar passare almeno 3 minuti prima
di effettuare una nuova misura sulla stessa persona.
Italiano

14

Misurazione della pressione
Misurazione con impostazione della pressione di gonfiaggio desiderata
Il UA-767S-W è stato progettato per rilevare in modo completamente automatico
la pressione e la frequenza cardiaca.
Applicare questo metodo se il gonfiaggio si ripete, o se i risultati non vengono
visualizzati malgrado la riduzione della pressione a 20 mmHg o meno.
1.

Avvolgere il bracciale sul braccio(preferibilmente il sinistro) all'altezzadel cuore.

A livello del cuore
Premere il pulsante START
START

2.

Premere il pulsante _START_ (AVVIO).

3.

Mentre lo zero lampeggia, premere etenere premuto il pulsante _START_(AVVIO) sino alla visualizzazione di
un valore superiore di 30 - 40 mmHg
rispetto alla pressione sistolica
prevista.
Premere e tenere premuto il START
pulsante START
Dopo aver raggiunto il valore
desiderato, rilasciare il pulsante_START_, la misurazione inizierà.
Proseguire quindi con la
misurazione della pressione
sanguigna come descritto alla
pagina precedente.

4.

Alla visualizzazionedella pressione digonfiaggio richiesta,
rilasciare il pulsantePer quanto riguarda la proceduradi rilevazione della pressione, fareriferimento alla pagina precedente

Note per una misurazione corretta

Sedersi in una posizione comoda. Appoggiare il braccio su un tavolo con il palmodella mano rivolto verso l’alto e con il bracciale alla stessa altezza del cuore.
Rilassarsi per circa 5 - 10 minuti prima di effettuare una misurazione. Se si èagitati o emozionati, la misurazione rifletterà questa situazione, con un valoredecisamente più alto (o più basso) della pressione arteriosa normale. Anche lafrequenza cardiaca sarà solitamente più alta del normale.
La pressione arteriosa di un soggetto varia costantemente, secondo l’attività fisica
ed emozionale, l’assunzione di cibi e la luce. Anche l’assunzione di bevande
alcoliche può avere un effetto sull’andamento dei valori di pressione arteriosa.
Questo apparecchio si basa sul principio di misura oscillometrico, quindi in gradodi valutare e analizzare i parametri delle pulsazioni cardiache. Se i battiti sono
molto deboli o irregolari, l’apparecchio potrebbe avere difficoltà a determinare lapressione arteriosa.
Se l'apparecchio rileva una circostanza anormale, arresterà la misurazione e
visualizzerà un simbolo di errore. (Vedere pagina 7 per la descrizione dei simboli).
Italiano15

Questo dispositivo deve essere gestito da persone adulte. Consultare il proprio
medico prima di utilizzare lo strumento su un bambino. In ogni caso, l’utilizzo su
bambini deve avvenire sempre sotto la stretta sorveglianza di persone adulte.
La prestazioni del monitor automatico per la pressione arteriosa potrebbero
essere influenzata dalla temperatura o umidità eccessiva o l'altitudine.

Richiamo delle misure memorizzate
Nota: Il dispositivo memorizza automaticamente gli ultimi 60 valori di pressione.
1.

Premere e tenere
Quando non è visualizzato alcun
premuto il pulsante START
elemento, premere e tenere premuto
START
il pulsante _START_ (AVVIO) perrichiamare i dati memorizzati.

2.

Rilasciare il pulsante quando si
visualizza la media dei dati.

3.

Vengono visualizzati
automaticamente il numero dei dati e
il numero dei dati memorizzati apartire dall'ultima misurazione.

4.

Il display si spegne automaticamente
dopo la visualizzazione di tutti i dati.

Media ditutti i dati

Pressione sistolica
media
Pressione diastolica
media
Frequenza cardiaca
media

Sistolica
Diastolica
Frequenza
cardiaca
Dati più recenti

Nota: se si preme il pulsante START(AVVIO) durante il richiamo dei dati,
il dispositivo si spegne.

Sistolica
Diastolica
Frequenza
cardiaca
Ultimi dati (Dati più vecchi)

Italiano

16

Cosa è l’indicatore IHB/AFib?
Quando il monitor rileva un ritmo irregolare durante le misurazioni, l’indicatore
IHB/AFib sarà visualizzato sul display unitamente ai valori di misurazione.
Nota: Raccomandiamo di contattare il proprio medico se viene indicato
frequentemente questo simbolo IHB/AFib.

Cosa è l’AFib?
Il cuore si contrae a causa di segnali elettrici che hanno luogo nel cuore e invia
sangue in tutto il corpo. La fibrillazione atriale (AFib) si verifica quando ilsegnale elettrico nell’atrio diventa confuso e causa disturbi nell’intervallo dellepulsazioni. L’AFib può causare ristagni del sangue nel cuore, i quali possono
causare facilmente coaguli di sangue, una delle cause di ictus e infarto.

％IHB/AFib%IHB/AFib è visualizzato come frequenza delI’IHB/AFib rilevato.
L’IHB/AFib può rilevare non soltanto i rumori come il movimento fisico, maanche un battito cardiaco irregolare. Si consiglia pertanto di consultare il
proprio medico se il livello di %IHB/AFib è elevato.
Numero di IHB/AFibrilevati in memoria
x 100 [%]
%IHB/AFib =Numero totale
Visualizzazione di %IHB/AFib: %IHB/AFib viene visualizzato quando si
visualizzano i valori medi. (Consultare “2.
Richiamo dei dati” in “Modalità operative”)%IHB/AFib non è visualizzato quando il numero
della memoria è inferiore o pari a sei.
Visualizzazione valori medi

Livello 0%IHB/AFib=0

Livello 1%IHB/AFib=1 - 9

Livello 2
Livello 3%IHB/AFib=10 - 24 %IHB/AFib=25 - 100

Non visualizzato

Italiano17

Indicatore dinamico della pressione
Lo strumento è provvisto di un indicatore dinamico a barre che controlla il
progresso della pressione durante la fase di misurazione. L’indicatore è visibile
nella parte sinistra del display e, si incrementa e diminuisce progressivamentedurante il gonfiaggio del bracciale e durante la misura della pressione.
Pressurizzazione

Depressurizzazione

Gonfiamento in corso

Misurazione in corso
Gonfiamento completato

Indicatore di classificazione dell’O.M.S.
L’Organizzazione Mondiale della Sanità (O.M.S.) hasancito delle linee guida per la corretta valutazione
e interpretazione dei valori di pressione misurata.
Ha suddiviso in 6 stadi la classificazione dei valori,come evidenziato dalla tabella sottostante:
Indicatore di classificazione O.M.S.
Ipertensione severa (grado 3)
Ipertensione moderata (grado 2)
Ipertensione lieve (grado 1)
Pressione normale – alta
Pressione normale
Pressione ottimale

Esempio:

Ipertensione moderata

Ipertensione lieve: L’indicatore mostra un segmento sulla basedei dati rilevati, corrispondente alla
Classifica zione O.M.S

Pressione normale – alta

Parlando di Pressione Arteriosa
Cosa si intende per Pressione Arteriosa?
La pressione arteriosa è la forza esercitata dal sangue contro le pareti delle arterie. La
pressione sistolica (massima) si verifica nel momento in cui il cuore si contrae e spinge ilsangue nelle arterie. La pressione diastolica (minima) si verifica quando il cuore si dilata,
ricevendo il sangue all’interno del ventricolo sinistro, e si prepara a una nuova
contrazione. La pressione arteriosa è misurata in millimetri di mercurio (mmHg).
Italiano18

Cosa si intende per Ipertensione Arteriosa?
Secondo le linee guida dell’Organizzazione Mondiale della Sanità, si può parlare diipertensione arteriosa quando si hanno valori di pressione sistolica e diastolica
persistentemente elevati oltre il limite di 140 / 90 mmHg. E’ importante però non fareiferimento esclusivamente ad una singola misura, ma è necessario considerare
anche il valore di più rilevazioni in un periodo di tempo stabilito dal medico ed anche
le fluttuazioni della pressione stessa.
Recenti studi a lungo termine hanno dimostrato che i limiti da considerare per lapressione domiciliare automisurata, sono infatti meno elevati e si attestano su valoridi 135 mmHg per la pressione sistolica e 85 mmHg per la pressione diastolica.

E come si controlla?
E’ possibile tenere sotto controllo l’ipertensione arteriosa variando il proprio stile divita e alimentare, evitando il più possibile situazioni di stress, oppure con opportuni
trattamenti farmacologici secondo prescrizione medica. Per prevenire l'ipertensione
arteriosa o per tenerla sotto controllo, le linee guida dell’ O.M.S. consigliano di:Non fumare
Eseguire esercizi a cadenza regolare
Ridurre l’uso di sale e moderare
Sottoporsi a check-up regolaril’assunzione di grassi animali
Mantenere il proprio peso a livelli ottimali

Perché è importante l’automisurazione al proprio domicilio?
La pressione che il medico misura mediante lo sfigmomanometro tradizionale è quasi
sempre caratterizzata da un rialzo generato dall’emozione che colpisce il paziente al
momento della visita medica, rialzo la cui entità e durata sono variabili da soggetto asoggetto. Questo fattore è comunemente conosciuto come ”ipertensione da camicebianco” o più opportunamente come “ipertensione clinica isolata” e si manifesta a
causa degli stati di ansia generati dalla presenza del medico o della struttura sanitaria.
Un vantaggio della misurazione domiciliare è che in genere è esente da questo
fenomeno. Un altro vantaggio dell’automisurazione è che permette di ottenere piùvalori nell’arco di diversi giorni o settimane, con una stima precisa delle variazionidella pressione.
La misurazione domiciliare inoltre può fornire utili informazioni nella valutazione deltrattamento antiipertensivo. In questo caso la pressione domiciliare può rivelarsi un
metodo molto efficace, perché il suo impiego coinvolge attivamente il paziente con
effetti potenzialmente positivi sull’aderenza alla terapia.

Classificazione della pressione secondo le linee guida “O.M.S.”
Gli standard per la valutazione della pressione
arteriosa, indipendentemente dall’età, sono
stati stabiliti dall’Organizzazione Mondiale della
Sanità (O.M.S.), come mostrato nella tabella afianco. Normalmente, la pressione arteriosa
aumenta durante l’attività quotidiana e il lavoroe, tende a diminuire durante il sonno. Le linee
guida inoltre consigliano di effettuarequotidianamente le misurazioni possibilmente sempre alla stessa ora.
Italiano19

L’esecuzione di molte rilevazioni permette diottenere un quadro della propria pressione
arteriosa più completo. E’ sempre meglio annotare
la data e l’ora della registrazione della pressione
arteriosa al fine di favorire una miglior diagnosi daparte del proprio medico curante.
Raccomandiamo di consultare sempre il proprio
medico al fine di interpretare correttamente i dati
relativi alla pressione.

Localizzazione guasti
ProblemaSul display, nonviene visualizzato
nessunmessaggio,
anche dopo
l’accensione

Causa probabile
Le batterie sono scariche.

Le batterie sono state inserite
non correttamente.
I poli delle batterie sono
posizionati in modo errato.
Il voltaggio delle batterie è
troppo basso. il simbolo
Il bracciale non si (BATTERIE SCARICHE) lampeggia.
gonfia.
In caso di batteriecompletamente scariche, ilsimbolo non sarà visualizzato.
Il bracciale non è avvolto inmodo adeguato.
Durante la fase di misurazione,
avete mosso il braccio o il
corpo.
Il dispositivo nonesegue lamisurazione.
La posizione del bracciale non
I valori ottenuti
è corretta.
sono
eccessivamentealti o bassi.

Vari

Il valore al domicilio è diverso
da quello misurato in clinica opresso lo studio medico.

Misura correttiva
Sostituire le batterie usate con
batterie nuove
Posizionare le batterie in modo
tale che i poli positivo e negativo
coincidano con quelli indicati nel
vano batterie.
Sostituire le batterie usate con
batterie nuove.
Avvolgere il bracciale in modo
adeguato.
Durante la fase di misurazione,
rimanere fermi e rilassati.
Sedere comodamente e rimanere
fermi e rilassati. Appoggiare il
braccio su un tavolo con il palmodella mano rivolto verso l’alto e
con il bracciale alla stessa altezzadel cuore.
Nel caso in cui il battito cardiaco
sia molto debole o irregolare, il
dispositivo potrebbe trovaredifficoltà nel determinare lapressione arteriosa.
Far riferimento al capitolo “Perchéè importante l’automisurazione al
proprio domicilio?”
Rimuovere le batterie. Posizionarlecorrettamente e eseguirenuovamente la misurazione.

Nota: Nel caso in cui le azioni sopra riportate non risolvano il problema, contattare il
rivenditore. Non tentare di aprire o riparare questo prodotto autonomamente,perché si rischia di invalidare la garanzia.
Italiano

20

Manutenzione
Non aprire il dispositivo che include componenti elettrici delicati ed un sofisticato
impianto ad aria che potrebbero danneggiarsi. Nel caso in cui non si riesce arisolvere il problema anche dopo aver eseguito quanto riportato nelle istruzioni dicui al capitolo “Localizzazione guasti”, contattare il centro di assistenza più vicino,
oppure rivolgersi al rivenditore o al punto vendita dal quale ha acquistato il
misuratore.
Il dispositivo è stato concepito e fabbricato per una lunga durata di servizio.
Tuttavia si raccomanda di controllare lo strumento ogni 2 anni, per garantire
funzionamento e precisione attendibili.

Dati tecnici
Tipo
Metodo di misurazione
Range di misurazione

Precisione
Alimentazione
Numero di misurazioni

Classificazione

Test clinici

EMD
Memoria dati
Condizioni d’esercizio
Condizioni di trasporto /conservazione
Dimensioni
Peso

UA-767S-W
Oscillometrico
Pressione: 0 - 299 mmHg
Pressione sistolica: 60-279 mmHg
Pressione diastolica: 40-200 mmHg
Pulsazioni: 40 - 180 battiti/minuto
Pressione: ±3 mmHg
Pulsazioni: ±5%
4 batterie da 1,5V (R6P, LR6 o AA) o
Alimentatore a rete elettrica (TB-233C) (non incluso)
Circa 700 LR6 (batterie alcaline)
Circa 200 R6P (batterie al manganese)
A una pressione di 180 mmHg, temperatura
ambiente di 23 °C.
Dispositivo medico elettronico ad alimentazione
interna (alimentazione con batterie) /Class II (alimentazione con alimentatore a rete)
Modalità di funzionamento continuo
In conformità ad ISO81060-2：2013
Nello studio di convalida clinica, K5 è stato utilizzato
su 85 soggetti per la determinazione della pressione
arteriosa diastolica.
IEC 60601-1-2: 2014
Le ultime 60 misurazioni
da +10° a +40°C/ da 15% a 85% RH /da 800 a 1060 hPa
da -20° a + 60°C /da 10% a 95% RH /da 700 a 1060 hPa
Circa 140 [L] x 60 [A] x 105 [P] mm
Circa 245 g, batterie escluse
Italiano21

Protezione contro
Parte applicata
Vita utile

Dispositivo: IP20
Bracciale Tipo BF
Dispositivo: 5 anni (se utilizzato sei volte al giorno)
Bracciale : 2 anni (se utilizzato sei volte al giorno)
Alimentatore: 5 anni (se utilizzato sei volte al giorno)

Alimentatore opzionale
L'alimentatore ha lo scopo di collegare il monitor della pressione sanguigna a una
fonte di alimentazione in ambiente domiciliare. Rivolgersi al rivenditore A&D di zona
per l'acquisto. L'alimentatore deve essere ispezionato o sostituito periodicamente.
Alimentatore opzionale

TB-233C

L'alimentatore ha lo scopo di collegare il monitor dellapressione sanguigna a una fonte di alimentazione inambiente domiciliare.
Per l’acquisto, contattare il punto vendita autorizzato
A&D.
L'alimentatore a rete elettrica deve essere ispezionato
o sostituito periodicamente.
Simboli riportati sull’alimentatore
SimboliFunzione / Significato
Per l'uso solo in ambienti interni
Dispositivo di Classe II
Fusibile termico
Fusibile
Etichetta del dispositivo relativa alla
direttiva CE
Etichetta del dispositivo relativa alla
certificazione EAC
Polarità dello spinotto dell'alimentatore

Accessori venduti separatamente
Bracciale Numero di catalogo Dimensione del bracciale
CUF-F-LA
CUF-I
CUF-F-A

Dimensione del braccio
Bracciale grande per adulti
31 cm a 45 cm
Bracciale ampio
22 cm a 42 cm
Bracciale per adulti
22 cm a 32 cm

Determinazione circonferenza del braccio: si misura a metà tra spalla e gomito.
Alimentatore a rete elettrica

Nota:

Numero di catalogo
TB-233C

Spina
Tipo CLe specifiche sono soggette a variazione, senza alcun preavviso.
La classificazione IP indica i gradi di protezione degli involucri in conformità con lanorma CEI 60529. Questo dispositivo è protetto contro corpi solidi estranei di 12mm di diametro o più grandi, come per esempio un dito. Questo dispositivo non è
protetto contro i getti d’acqua.

Italiano22Índice
Estimados clientes ............................................................................ 2
Observações preliminares ................................................................ 2
Precauções ....................................................................................... 2
Identificação de componentes .......................................................... 5
Símbolos ........................................................................................... 6
Modo de funcionamento ................................................................... 8
1. Medição normal ....................................................................... 8
2. Revisão dos dados da memória .............................................. 8
3. Eliminação de todos os dados armazenados na memória...... 8
4. Medição com a tensão sistólica desejada ............................... 8
Utilização do dispositivo de monitorização ....................................... 9
Instalação/substituição das pilhas ............................................... 9
Ligação do tubo de ar ............................................................... 10
Ligação do adaptador de CA .................................................... 10
Seleção do tamanho correto da braçadeira .............................. 11
Aplicação da braçadeira ............................................................ 11
Como efetuar medições precisas .............................................. 13
Medição..................................................................................... 13
Depois da medição ................................................................... 13
Medições ........................................................................................ 14
Medição normal ......................................................................... 14
Medição com a tensão sistólica desejada ................................. 15
Notas para medições corretas .................................................. 15
Revisão dos dados da memória ..................................................... 16
O que é o Indicador de IHB/FibA? .................................................. 17
A que se refere FibA? ..................................................................... 17%IHB/FibA ...................................................................................... 17
Indicador da barra de tensão .......................................................... 18
Indicador de classificação segundo a OMS .................................... 18
Sobre tensão arterial ...................................................................... 18
O que é a tensão arterial? ......................................................... 18
O que é a hipertensão e como se controla? ............................. 18
Porque deve medir a tensão arterial em casa?......................... 19
Classificação da tensão arterial segundo a OMS ..................... 19
Variações da tensão arterial ..................................................... 19
Resolução de problemas ................................................................ 20
Manutenção .................................................................................... 21
Dados técnicos ............................................................................... 21

Português 1

Estimados clientes
Parabéns pela aquisição de um dispositivo de monitorização de tensão
arterial A&D de última geração. Desenvolvido para facilidade deutilização e precisão, este dispositivo irá facilitar a sua rotina de diária na
medição da tensão arterial.
Recomendamos que leia atentamente o presente manual de
instruções antes de utilizar o dispositivo pela primeira vez.

Observações preliminares
Este dispositivo encontra-se em conformidade com a Diretiva Europeia
93/42 CEE para produtos médicos. Tal é evidenciado pela marca deconformidade. (0123: O número de referência ao organismo
notificado envolvido)
O dispositivo foi desenvolvido para utilização em adultos, não devendo
ser utilizado em recém-nascidos ou bebés.
Âmbito de utilização. O dispositivo destina-se à utilização autónoma porparte do utilizador para cuidados de saúde em ambiente doméstico.
Este dispositivo foi desenvolvido para medir a tensão arterial e afrequência de pulso de pessoas para fins de diagnóstico.

Precauções
A construção deste dispositivo inclui componentes de precisão. Devem
evitar-se extremos de temperatura, humidade, luz solar direta, choque ou
poeira.
Limpe o dispositivo e a braçadeira com um pano seco e macio ou com
um pano humedecido com água e um detergente neutro. Nunca utilizeálcool, gasolina, diluente ou outros produtos químicos agressivos para
limpar o dispositivo ou a braçadeira.
Evite dobrar a braçadeira apertada ou guardar o tubo torcido por
períodos prolongados, uma vez que tal pode encurtar a vida útil dos
componentes.
Tenha cuidado para evitar o estrangulamento acidental de bebés ou
crianças com o tubo e o cabo.
Não torça o tubo de ar durante a medição. Caso contrário, podem
resultar lesões devido à tensão contínua da braçadeira.
O dispositivo e a braçadeira não são resistentes à água. Evite que
a chuva, a transpiração e a água molhem o dispositivo e a braçadeira.
As medições podem sofrer distorções se o dispositivo for utilizado próximo
de televisores, micro-ondas, telemóveis, raios-X ou outros dispositivos
com fortes campos elétricos.
Os dispositivos de comunicação sem fios, tais como os dispositivos de rededoméstica, telemóveis, telefones sem fios e as respetivas centrais ewalkie-talkies, podem afetar o dispositivo de monitorização de tensão
arterial.
Como tal, é necessário manter uma distância de 30 cm entre o dispositivo
de monitorização e os referidos dispositivos.
Português 2

Quando reutilizar o dispositivo, confirme se está limpo.
O dispositivo, os respetivos componentes e as pilhas usados não são
tratados como resíduos domésticos comuns e devem ser eliminados deacordo com os regulamentos locais aplicáveis.
Quando utilizar o adaptador de CA, assegure que o adaptador de CA
pode ser prontamente retirado da tomada elétrica quando necessário.
Não modifique o dispositivo. A eventual modificação do dispositivo podeprovocar acidentes ou danos neste.
Para medir a tensão arterial, deve apertar o braço com a braçadeira osuficiente para deter temporariamente o fluxo sanguíneo através da artéria.
Tal pode provocar dor, dormência ou uma marca vermelha temporária no
braço. Este estado aparece especialmente quando a medição se repete
sucessivamente. Qualquer dor, dormência ou marcas vermelhas
desaparecem com o tempo.
Se medir a tensão arterial com demasiada frequência, tal pode resultar em
danos devido à interferência com o fluxo sanguíneo. Se utilizar o dispositivo
de forma repetida, verifique se a utilização deste não resulta numa
disfunção prolongada da circulação sanguínea.
Caso tenha sido submetido a uma mastectomia, consulte o seu médico antes
de utilizar o dispositivo.
Não deixe que as crianças utilizem o dispositivo sem a supervisão de um
adulto, nem utilize o dispositivo num local dentro do alcance de crianças.
Tal pode provocar acidentes ou danos.
O dispositivo contém peças pequenas que podem provocar asfixia caso as
crianças as ingiram acidentalmente.
Retire o adaptador de CA do dispositivo quando não estiver a utilizá-lo
durante a medição.
O uso de acessórios não indicados neste manual pode comprometer asegurança.
Caso a bateria produza um curto-circuito, esta pode ficar quente e
eventualmente causar queimaduras.
Permita que o dispositivo se adapte ao ambiente circundante antes deutilizá-lo (cerca de uma hora).
Não foram realizados ensaios clínicos em recém-nascidos e em mulheres
grávidas. Não usar em recém-nascidos ou em mulheres grávidas.
Não toque simultaneamente nas pilhas, na tomada de CC e no paciente.
Caso contrário, tal pode resultar numa descarga elétrica.
Não encha sem colocar a braçadeira no antebraço.

Contraindicações

Em seguida, indicam-se as precauções para o uso correto do dispositivo.
Não aplique a braçadeira num braço com outro dispositivo elétrico
médico conectado. O dispositivo pode não funcionar corretamente.
As pessoas com um grave défice circulatório no braço devem consultar um
médico antes de utilizar o dispositivo, de modo a evitar problemas médicos.
Não autodiagnostique os resultados de medição nem inicie o tratamento
por iniciativa própria. Consulte sempre o seu médico para a avaliação
dos resultados e tratamento.
Não aplique a braçadeira num braço com uma ferida por curar.
Não aplique a braçadeira num braço que está a receber tratamento por
Português 3

via intravenosa ou transfusão sanguínea. Tal pode provocar lesões ou
acidentes.
Não utilize o dispositivo na presença de gases inflamáveis como, por
exemplo, gases anestésicos. Tal pode provocar uma explosão.
Não utilize o dispositivo em ambientes com alta concentração deoxigénio, tais como as câmaras de oxigénio de alta pressão ou tenda deoxigénio. Tal pode provocar um incêndio ou explosão.

Português 4

Identificação de componentes
Ecrã

Ficha adaptadorade CA

Ficha do conector de ar

Tomada de CCBotão START (Iniciar)
Bocal de ar

Tubo de ar

Extensão de colocação

Marca de posição da artéria
Marca de índice
BraçadeiraCompartimento das pilhas
Tampa das pilhas

Pilhas de 1,5 V(R6P, LR6 ou AA)

Ecrã

Média
MEMÓRIA

Símbolo de IHB/FibA
Tensãosistólica%IHB/FibA
Símbolo de erro de ajuste da braçadeiraTensãodiastólica

Símbolo de erro por movimento
Marca de coração

Frequência depulso

Indicador de classificação segundo a
OMS e indicador da barra de tensão
Indicação de pilhas

Português 5

Símbolos
Símbolos impressos na caixa do dispositivo
Símbolos
SYS
DIA
PUL

IP

Função/significado
Dispositivo em modo de espera e ligar o dispositivo.
Tensão arterial sistólica em mmHg
Tensão arterial diastólica em mmHg
Pulsação por minuto
Guia de instalação das pilhas
Corrente contínua
Tipo BF: o dispositivo, a braçadeira e o tubo foram
desenvolvidos para proporcionar proteção especial contra
choques elétricos.
Etiqueta de dispositivos médicos segundo diretiva de CE
Representante da UE
FabricanteData de fabrico
Símbolo de proteção internacional
Etiqueta REEE

SN

Número de série
Consulte o manual/folheto de instruções
Polaridade da tomada de CC
Manter seco

Símbolos que são apresentados no ecrãSímbolosFunção/significado
Medida recomendada
É apresentado quando a medição A medição está em curso.
está em curso. Este indicador pisca Permanecer o mais imóvel
quando a pulsação é detetada.
possível.
Símbolo de IHB/FibA
É apresentado quando se deteta
um batimento cardíaco irregular.
Pode acender quando se deteta
uma vibração muito ligeira comotremor ou agitação.
A leitura pode produzir um
valor incorreto. Realize a
Aparece quando se deteta um
medição novamente.
movimento do corpo ou do braço.
Permaneça imóvel durante
a medição.
Português 6

Símbolos
SímbolosFunção/significado

Medida recomendada
A leitura pode produzir um
É apresentado durante a medição valor incorreto. Coloque aquando a braçadeira não está bem braçadeira corretamente e
ajustada
realize a medição
novamente.
Índice detetado de IHB/FibA na
memória%IHB/FibA =Número de IHB/FibsA
detetados na memória x100 [%]Número total

Medições anteriores armazenadas
na MEMÓRIA.
Dados médios
PILHAS CARREGADA
Indicador da potência das pilhas
durante a medição.
PILHAS GASTAS
Substitua todas as pilhas por
As pilhas estão fracas quando esta novas quando a marca
marca pisca.
piscar.
Realize a medição
novamente.
Tensão arterial instável devido amovimento durante a medição.
Permaneça totalmente
imóvel durante a medição.
Os valores sistólicos e diastólicos
têm um intervalo máximo de
10 mmHg entre si.
O valor da tensão não aumentou
durante a inflação.
Coloque a braçadeiracorretamente e realize a
A braçadeira não está colocada
medição novamente.
corretamente.
ERRO DE VISUALIZAÇÃO DA
PULSAÇÃO
A pulsação não foi detetada
corretamente.
Remova as pilhas, pressioneo botão START e, em
Erro interno do dispositivo deseguida, volte a colocar as
monitorização de tensão arterial
pilhas. Se o erro continuar
a aparecer, contacte o seu
distribuidor.
Português 7

Modo de funcionamento
1. Medição normal
Pressione o botão START. É medida a tensão arterial e os dados são
armazenados na memória. Este aparato pode armazenar as últimas 60
medições na memória.

2. Revisão dos dados da memória
Quando não visualizar quaisquer dados,pressione e mantenha o botão START
pressionado.

No modo de espera,
pressione e mantenha ootãopressionado
START

Solte o botão quando visualizar os dados
médios.

O número de dados e os dados
armazenados são visualizados
automaticamente por ordem desde a última
medição.

Para mais informações sobre a revisão dedados, consulte “Revisão dos dados damemória”.

3. Eliminação de todos os dados armazenados
na memória
Quando desligar o dispositivo, pressione e mantenha
o botão START pressionado até visualizar “”Selecione “” para apagar os dados.
Os dados são apagados quando a marca
piscar.
O dispositivo desliga automaticamente.

START

4. Medição com a tensão sistólica desejada
Consulte a página 15 para a medição com a tensão sistólica desejada.
Português 8

Utilização do dispositivo de monitorização
Instalação/substituição das pilhas
1. Retire a tampa das pilhas.

Passo 1
Passo 3

2. Retire as pilhas usadas e insira aspilhas novas no compartimento daspilhas, tal como descrito,
respeitando as polaridades (+ e -)das pilhas.
Utilize apenas pilhas R6P, LR6 ou
AA.

Passo 1

Passo 2

3. Coloque a tampa das pilhas.

CUIDADOInsira as pilhas no compartimento das pilhas, tal como ilustrado na figura.
Caso contrário, o dispositivo não funcionará.
Quando a marca(marca de PILHAS GASTAS) piscar no ecrã,substitua todas as pilhas por pilhas novas. Não misture pilhas usadas com
pilhas novas. Tal pode encurtar a vida útil das pilhas ou causar o mau
funcionamento do dispositivo.
Substitua as pilhas pelo menos dois segundos após o dispositivo desligar.
Se a marca(marca de PILHAS GASTAS) aparecer até mesmo
depois de substituir as pilhas, realize uma medição da tensão arterial. O
dispositivo pode então reconhecer as pilhas novas.
A marca (marca de PILHAS GASTAS) não aparece quando as pilhasestão totalmente descarregadas.
A vida útil das pilhas varia com a temperatura ambiente e pode serreduzida com temperaturas baixas. Em geral, quatro pilhas LR6 novas
duram aproximadamente um ano quando utilizadas para realizar
medições duas vezes por dia.
Utilize unicamente as pilhas especificadas. As pilhas fornecidas com odispositivo são para testar o funcionamento do dispositivo demonitorização e podem ter uma vida útil mais curta.
Retire as pilhas quando não pretender utilizar o dispositivo durante um
período prolongado.
As pilhas podem ter fugas e causar um mau funcionamento.

Português 9

Utilização do dispositivo de monitorização
Ligação do tubo de ar

Bocal de arBocal de ar

Insira firmemente a ficha doconector de ar no bocal de ar.

Ficha do conector de ar

Ligação do adaptador de CA
Insira a ficha do adaptador de CA
na tomada de CC.
Em seguida, ligue o adaptador deCA à tomada elétrica.

Tomada de CC

Ficha adaptadora de CA
Utilize o adaptador de CA
especificado. (Consulte apágina 22.)
Quando desligar o adaptador de CA da tomada elétrica, agarre e puxe
o corpo do adaptador de CA para fora da tomada.
Quando desligar a ficha adaptadora de CA do dispositivo demonitorização de tensão arterial, agarre e puxe a ficha adaptadora de
CA para fora do dispositivo de monitorização.

Português 10

Utilização do dispositivo de monitorização
Seleção do tamanho correto da braçadeira
A seleção de uma braçadeira de tamanho correto é importante para uma
leitura correta. Se a braçadeira não for do tamanho correto, a leitura podeindicar um valor incorreto da tensão arterial.
O tamanho do braço está impresso em cada braçadeira.
O índicee a gama adequada de colocação da braçadeira indicam-lhe seestá a colocar a braçadeira corretamente. (Consulte os “Símbolos impressos
na braçadeira” na página seguinte)
Se o índice
se encontra fora da gama, entre em contacto com o
distribuidor local para adquirir uma braçadeira de substituição.
A braçadeira é um produto consumível. Caso esta se desgaste, adquirauma nova.
Número decatálogo
CUF-F-LA
CUF-I
CUF-F-A

Tamanho do braço Tamanho da braçadeira recomendado

31 cm a 45 cm
Braçadeira grande para adultos
22 cm a 42 cm
Braçadeira de grande extensão
22 cm a 32 cm
Braçadeira para adultos
Tamanho do braço: A circunferência no bíceps.
Nota:
O UA-767S-W não foi desenvolvido para utilização
de uma braçadeira pequena.

Aplicação da braçadeira
1. Coloque a braçadeira em redor dobraço, uns 1 - 2 cm acima da parteinterior do cotovelo, tal como
ilustrado na figura.
Coloque a braçadeira em contacto
direto com a pele, dado que a roupa
pode causar uma pulsação fraca e
provocar erro de medição.

Evite deixar a manga demasiado apertada ao
enrolá-la para cima.
Braçadeira1 - 2 cm

Tubo de arMarca de posição da artéria

2. A constrição do braço, causada ao
enrolar a manga apertada, pode
impedir leituras precisas.
3. Confirme se o índiceseencontra dentro da gama adequadade colocação.

Português 11

Aperte firmemente ofixador de tecido.

Índicee gama adequada de colocação

Utilização do dispositivo de monitorização
Símbolos impressos na braçadeiraFunção/significado

SímbolosMedida recomendada
Ajuste a marcanaartéria do antebraço ou deforma alinhada com odedo anelar na parteinterior do braço.

Marca de posição da artéria▲
REFÍndiceNúmero de catálogo
Gama adequada de ajuste dabraçadeira para adultos.
Está impressa na braçadeira para
adultos.

Utilize a braçadeira grande
Acima da gama impressa na
para adultos em vez dabraçadeira para adultos e braçadeira braçadeira para adultos ou
de grande extensão.
a braçadeira de grande
extensão.
Gama adequada de ajuste dabraçadeira de grande extensão. Está
impressa na braçadeira de grande
extensão.
Gama adequada de ajuste dabraçadeira grande para adultos. Está
impressa na braçadeira grande para
adultos.
Abaixo da gama impressa na
braçadeira para adultos e braçadeira
de grande extensão.
Utilize a braçadeira para
Abaixo da gama impressa na
adultos em vez dabraçadeira grande para adultos.
braçadeira grande para
adultos.
Número de loteBraçadeira grande para adultos

Gama adequada de ajusteA

L

Braçadeira de grande extensão

S

W

L

S

A

L

Braçadeira para adultos

Português 12

Utilização do dispositivo de monitorização
Como efetuar medições precisas
Tenha em conta os seguintes aspetos para efetuar a medição da tensão
arterial com a máxima precisão:Sente-se comodamente numa cadeira. Apoie o seu braço em cima da
mesa. Não cruze as pernas. Mantenha os pés assentes no chão eendireite as costas.
Relaxe durante cerca de cinco a dez minutos antes de efetuar a medição.
Coloque o centro da braçadeira ao mesmo nível que o seu coração.
Permaneça quieto e em silêncio durante a medição.
Não efetue a medição imediatamente após praticar exercício físico ou
tomar banho. Descanse cerca de vinte ou trinta minutos antes de efetuar amedição.
Tente medir a sua tensão arterial à mesma hora todos os dias.

Medição
Durante a medição, é normal sentir a braçadeira muito apertada. (Não sealarme por isso).

Depois da medição
Após a medição, pressione o botão START para desligar o dispositivo.
Retire a braçadeira e registe os seus dados.
Nota: O dispositivo possui uma função de desconexão automática daalimentação que desliga o dispositivo aproximadamente um minuto após amedição.
Deixe decorrer pelo menos três minutos entre medições na mesma
pessoa.

Português 13

Medições
Antes de efetuar a medição, consulte “Notas para medições corretas” napágina seguinte.

Medição normal
1.

Coloque a braçadeira no seu braço
Ao nível do coração
(preferencialmente o braço esquerdo).
Sente-se tranquilamente durante a medição.
2. Pressione o botão START.
Pressione START
Visualiza todos os segmentos do ecrã.
É apresentado zero (0) a piscar
brevemente.
Visualiza todos os
A indicação no ecrã muda, conforme
segmentos do ecrãindicado na figura à direita, no início damedição. A braçadeira começa a encher.
É normal sentir a braçadeira muito
apertada. É visualizado o indicador dabarra de tensão durante a inflação no
É apresentado
canto esquerdo do ecrã.
zero
Iniciaainflação
Nota: Se desejar parar a inflação a qualquer
altura, pressione novamente o botão
START.
3. Quando a inflação estiver concluída, adeflação começa automaticamente e a
Pressurização
marca(marca do coração) pisca,
indicando que a medição está em curso.
Uma vez detetada pulsação, a marca
pisca com cada batimento do coração.
Nota: Se não obtiver uma tensão
apropriada, o dispositivo começa
novamente a encher
automaticamente.
Para evitar uma nova inflação,consulte “Medição com a tensão
A medição
sistólica desejada” na página
está em curso
seguinte.
4. Quando a medição estiver concluída, sãovisualizadas as leituras das tensões
Tensão sistólica
sistólica e diastólica e a frequência depulso.
Tensão diastólica
A braçadeira expulsa o ar residual e Classificação segundo a OMS
esvazia completamente.
Frequência da pulsação
5. Pressione o botão START para desligaro dispositivo.
Esvazia
Após um minuto de inatividade, oautomaticamente odispositivo desliga automaticamente.
ar residual
Nota: Deixe decorrer pelo menos três minutos
entre medições na mesma pessoa.
Português 14

Medições
Medição com a tensão sistólica desejada
O UA-767S-W foi desenvolvido para detetar a pulsação, enchendo
automaticamente a braçadeira até ao nível de tensão sistólica.
Utilize este método quando a repetição da inflação ocorrer ou quando os
resultados não forem visualizados, mesmo no caso em que a tensão
diminua para 20 mmHg ou menos.
1.

Coloque a braçadeira no braço ao nível do
Ao nível do coração
coração (preferencialmente o braçoesquerdo).
Pressione o botão START

2.

Pressione o botão START.

3.

Quando o zero piscar, pressione e
mantenha o botão START
pressionado até aparecer um número
cerca de 30 a 40 mmHg superior à
sua tensão sistólica prevista.

4.

Quando o número desejado foratingido, solte o botão START para
iniciar a medição. Continue a medir asua tensão arterial conforme descritona página anterior.

Solte o botão natensão sistólica
desejada

Pressione e
mantenha obotãopressionado

START

Consulte a página anteriorpara a medição
Notas para medições corretas
Sente-se numa posição cómoda. Coloque o braço em cima de uma mesa
com a palma virada para cima e a braçadeira ao mesmo nível que ocoração.
Relaxe durante cerca de cinco a dez minutos antes de efetuar a medição.
Se estiver agitado ou deprimido devido a stress emocional, a medição irá
refletir este stress como uma leitura mais alta (ou mais baixa) do que aleitura da tensão arterial normal e a leitura da pulsação será geralmentemais rápida do que a normal.
A tensão arterial de uma pessoa varia constantemente, dependendo
daquilo que estiver a fazer e daquilo que comeu. Aquilo que bebe podeter um efeito muito forte e rápido sobre a sua tensão arterial.
Este dispositivo baseia as medições no batimento cardíaco. Se tiver um
batimento cardíaco muito fraco ou irregular, este dispositivo pode ter
dificuldade em determinar a sua tensão arterial.
Português 15

No caso de o dispositivo detetar uma condição anormal, o mesmo
interrompe a medição e indica um símbolo de erro no visor. Consulte apágina 7 para a descrição dos símbolos.
Este dispositivo destina-se à utilização em adultos. Consulte o seu
médico antes de utilizar este dispositivo numa criança. Uma criança não
deve utilizar este dispositivo sem a ajuda de um adulto.
O desempenho do dispositivo de monitorização de tensão arterial
automático pode ser afetado por temperatura ou humidade excessivas
ou pela altitude.

Revisão dos dados da memória
Nota: Este dispositivo armazena as últimas 60 medições na memória.
1.

Quando não forem visualizados
dados, pressione e mantenha obotão START pressionado pararever os dados armazenados.

Pressione e
mantenhapressionado

START

botão
Tensão sistólica médiaTensão diastólica média
Frequência média dapulsação

2.

Solte o botão quando visualizar os
dados médios.

3.

O número de dados e os dados
armazenados são visualizados
automaticamente por ordem desdea última medição.

4.

O ecrã desliga automaticamente
após todos os dados serem
visualizados.

Média detodos os dados

Sistólica
Diastólica
Frequência da pulsação
Dados mais recentes

Nota: Se pressionar o botão START
enquanto revê os dados, odispositivo desliga.

Sistólica
Diastólica
Frequência da pulsação
Últimos dados (mais antigos)

Português 16

O que é o Indicador de IHB/FibA?
Se o dispositivo de monitorização detetar um ritmo irregular durante asmedições, aparece o indicador de IHB/FibA no visor com os valores demedição.
Nota: Recomendamos que consulte o seu médico se visualizar este
indicador de IHB/FibA

com frequência.

A que se refere FibA?
O coração contrai devido a correntes elétricas que ocorrem no coração eenvia o sangue por todo o corpo. A fibrilação atrial (FibA) ocorre quando ascorrente elétrica do átrio cardíaco se confundem e originam alterações nos
intervalos de pulsação. A FibA pode ocasionar estagnações de sangue nocoração, as quais podem produzir facilmente coágulos de sangue, uma dascausas de AVC e de enfartes cardíacos.

%IHB/FibA%IHB/FibA é visualizado como frequência de IHB/FibA detetado.
IHB/FibA pode detetar não só interferências tais como o movimento físico,mas também um batimento cardíaco irregular. Como tal, recomendamos
que consulte o seu médico se o %IHB/FibA é elevado.
Número de IHB/FibsA
detetados na memória%IHB/FibA =x 100 [%]

Número total

Visualização de %IHB/FibA: %IHB/FibA é visualizado quando são
visualizados os valores médios.
(Consulte o “2. Revisão dos dados damemória” em “Modo de funcionamento”)%IHB/FibA não é visualizado quando onúmero de memória for igual a seis ou inferior.
Visualização do valor médio

Nível 0
%IHB/FibA=0

Nível 1
%IHB/FibA=1 - 9

Nível 2
%IHB/FibA=10 - 24

Não visualizado

Português 17

Nível 3
%IHB/FibA=25 - 100

Indicador da barra de tensão
O indicador monitoriza o progresso da tensão durante a medição.
Pressurização

Libertação do ar

Inflação em curso

Medição em curso
Inflação completa

Indicador de classificação segundo a OMS
Cada um dos segmentos do indicador debarra corresponde à classificação datensão arterial segundo a OMS descrita napágina seguinte.

Exemplo:

Indicador de classificação segundo a OMS

Hipertensão grave
Hipertensão moderada
Hipertensão leve
Alta normal
Normal
Ideal

Hipertensão moderada

Hipertensão leve

: O indicador visualiza um segmento baseado
nos dados atuais, correspondentes àclassificação segundo a OMS.

Alta normal

Sobre tensão arterial
O que é a tensão arterial?
A tensão arterial é a força que o sangue exerce contra as paredes dasartérias. A tensão sistólica ocorre quando o coração contrai. A tensão
diastólica ocorre quando o coração expande. A tensão arterial é medida em
milímetros de mercúrio (mmHg). A tensão arterial natural de uma pessoa é
representada pela tensão fundamental, a qual se mede logo pela manhã
quando a pessoa ainda está em repouso e antes de se alimentar.

O que é a hipertensão e como se controla?

A hipertensão é a tensão arterial anormalmente alta e, caso não seja
corrigida, pode causar muitos problemas de saúde, incluindo AVC e
Português 18

enfarte cardíaco. A hipertensão pode ser controlada através da alteração
do estilo de vida, evitando o stress e com medicamentos mediante a
supervisão de um médico.
Para evitar a hipertensão ou para mantê-la sob controlo:Não fume
Faça exercício regularmente
Reduza a ingestão de sal ede gordura
Faça exames físicos regulares
Mantenha o peso ideal

Porque deve medir a tensão arterial em casa?

A medição da tensão arterial numa clínica ou num consultório médico podecausar apreensão e, por isso, produzir uma leitura elevada, 25 a 30 mmHg
superior à medida em casa. A medição em casa reduz os efeitos deinfluências externas sobre as leituras da tensão arterial, complementa as
leituras do médico e proporciona um historial mais completo e exato datensão arterial.

Classificação da tensão arterial segundo a OMS

Variações da tensão
arterial

Material de referência; Journal of Hypertension 1999, Vol 17

Hipertensão de grau 3 (grave)Tensão arterial diastólica

A Organização Mundial de
Saúde (OMS) estabeleceu
as normas para a avaliação
da tensão arterial,
independentemente daidade, conforme ilustrado
no gráfico.

Hipertensão de grau 2 (moderada)
Hipertensão de grau 1 (leve)
Alta-normal
NormalIdealTensão arterial

A tensão arterial de umaTensão arterial sistólica
pessoa variaconsideravelmente numa base diária e sazonal. Esta pode variar entre
30 a 50 mmHg dependendo de diversas condições durante o dia. Nas
pessoas hipertensas, estas variações são ainda mais notáveis.
Normalmente, a tensão arterial sobe quando a pessoa está a trabalhar
ou a jogar e baixa para os seus níveis mais baixos durante o sono.
Portanto, não se preocupe demasiado com os resultados de uma única
medição.
Variação típica num dia
Efetue as medições à mesma hora todos(Medição a cada cinco minutos)
Sistólica
os dias, utilizando o procedimento
descrito neste manual e familiarize-se
com a sua tensão arterial normal. As
leituras regulares proporcionam um
historial da tensão arterial mais completo.
Assegure que anota a data e a hora ao
Diastólica
DormirHoraregistar a sua tensão arterial. Consulte o
seu médico para interpretar os dados dasua tensão arterial.

Português 19

Resolução de problemas
ProblemaNão são
visualizados
quaisquer
dados no visor,
mesmo quando
o dispositivo
está ligado.

A braçadeiranão enche.

Causa provável
As pilhas estão gastas.
Os terminais das pilhasnão estão na posição
correta.
A tensão das pilhas é
demasiado baixa.
A marca (marca de
PILHAS GASTAS) pisca.
Se as pilhas estiverem
completamente gastas, amarca não aparece.
A braçadeira não está
colocada
adequadamente.

Medida recomendada
Substitua todas as pilhas por
novas.
Volte a colocar as pilhas com os
terminais negativos e positivos
em conformidade com os
terminais indicados nocompartimento das pilhas.

Substitua todas as pilhas por
novas.

Aplique a braçadeiracorretamente.

Assegure que permanece
imóvel e calmo durante a
O dispositivo
medição.
não efetua
medições.
Sente-se comodamente e imóvel.
As leituras são
Coloque o braço em cima de uma
A posição da braçadeira
demasiado
mesa com a palma virada paranão é a correta.
altas ou
cima e a braçadeira ao mesmo
demasiado
nível que o coração.
baixas.
Se tiver um batimento cardíaco
muito fraco ou irregular, odispositivo pode ter dificuldade
em determinar a sua tensão
arterial.
O valor é diferente doConsulte “Porque deve medir avalor medido numa
clínica ou numtensão arterial em casa?”.
consultório médico.
Outro
Retire as pilhas. Volte a
colocá-las corretamente e tente
efetuar a medição novamente.
Nota: Se as medidas descritas anteriormente não solucionarem o problema,contacte o distribuidor. Não tente abrir ou reparar este produto, dado
que qualquer tentativa nesse sentido invalida a sua garantia.
Deslocou o seu braço ou
corpo durante a medição.

Português 20

Manutenção
Não abra o dispositivo. Este contém componentes elétricos delicados e
uma unidade de ar complexa que podem sofrer danos. Caso não
solucione o problema através das instruções referentes à resolução deproblemas, contacte o distribuidor autorizado na sua área de residência ou
o departamento de apoio ao cliente da A&D. O serviço de apoio ao cliente
da A&D fornece informações técnicas, peças sobresselentes e unidades
aos distribuidores autorizados.
O dispositivo foi desenvolvido e fabricado para atingir uma vida útilprolongada. Não obstante, recomenda-se geralmente que o dispositivo seja
submetido a uma inspeção a cada 2 anos a fim de assegurar ofuncionamento e a precisão adequados. Contacte o distribuidor autorizado
na sua área de residência ou a A&D para efeitos de manutenção.

Dados técnicos
Tipo
Método de medição
Intervalo de medição

Precisão de medição
Fonte de alimentação
Número de medições

Classificação

Ensaio clínico

EMD
Memória
Condições defuncionamento
Condições de transporte/ armazenamento
Dimensões
Peso

UA-767S-W
Medição oscilométricaTensão: 0 - 299 mmHg
Tensão sistólica: 60 - 279 mmHg
Tensão diastólica: 40 - 200 mmHg
Pulsação: 40 - 180 batimentos/minutoTensão: ±3 mmHg
Pulsação: ±5%
4 x pilhas 1,5 V (R6P, LR6 ou AA) ou
adaptador de CA (TB-233C) (Não incluído)
Aprox. 700 vezes
LR6 (pilhas alcalinas)
Aprox. 200 vezes
R6P (pilhas de manganês)Com um valor de tensão de 180 mmHg,
temperatura ambiente de 23 °C.
Equipamento ME alimentado internamente(alimentado por pilhas) / Classe II (fornecido por
adaptador)
Modo de funcionamento contínuo
De acordo com a ISO81060-2：2013
No estudo de validação clínica, foi utilizado K5 em
85 doentes para determinação da tensão arterial
diastólica.
IEC 60601-1-2: 2014Últimas 60 medições
+10 a +40 °C / 15 a 85%HR / 800 a 1.060 hPa-20 a +60 °C / 10 a 95%HR / 700 a 1.060 hPa
Aprox. 140 [L] x 60 [A] x 105 [P] mm
Aprox. 245 g, excluindo as pilhas
Português 21

Proteção

Dispositivo:

IP20

Componente aplicado
Vida útil

Braçadeira Tipo BF
Dispositivo: 5 anos (quando utilizado seis vezes
por dia)
Braçadeira: 2 anos (quando utilizado seis vezes
por dia)
Adaptador de CA: 5 anos (quando utilizado seisvezes por dia)
Adaptador de CA acessório
O adaptador serve para ligar o dispositivo a uma fonte de alimentação em
casa. Contacte o seu distribuidor A&D local para efetuar a aquisição.
O adaptador de CA deve ser inspecionado ou substituído periodicamente.
Contacte o seu distribuidor A&D local para efetuar
TB-233Ca aquisição.
O adaptador de CA deve ser inspecionado ou
substituído periodicamente.
Símbolos impressos no adaptador de CA
SímbolosFunção/significado
Exclusivamente para uso interno
Dispositivo de classe II
Fusível térmico
FusívelEtiqueta de aparelhos segundo
diretiva de CE
Etiqueta de aparelhos segundo acertificação EAC
Polaridade da ficha adaptadora de CA
Acessórios vendidos separadamente
Número deBraçadeira
Tamanho da braçadeira
Tamanho do braço
catálogo
CUF-F-LA Braçadeira grande para adultos 31 cm a 45 cm
CUF-I
Braçadeira de grande extensão 22 cm a 42 cm
CUF-F-A
Braçadeira para adultos
22 cm a 32 cm
Tamanho do braço: A circunferência no bíceps.
Conector (tipo deAdaptador de CA
Número de catálogo
tomada)TB-233C
Tipo C
Nota: As especificações estão sujeitas a alteração sem aviso prévio.
A classificação IP corresponde aos graus de proteção fornecidos
pelas caixas de acordo com a IEC 60529. Este dispositivo está
protegido contra objetos estranhos sólidos com 12 mm de diâmetro
e maiores do que os dedos, pwor exemplo. Este dispositivo não está
protegido contra água.
Português 22

Inhoud
Beste klant ........................................................................................ 2
Inleidende opmerkingen ................................................................... 2
Voorzorgsmaatregelen ..................................................................... 2
Identificatie onderdelen..................................................................... 5
Symbolen .......................................................................................... 6
Gebruiksmodus................................................................................. 8
1. Normale meting........................................................................................... 8
2. De gegevens opvragen.............................................................................. 8
3. Alle gegevens in het geheugen wissen.................................................... 8
4. Meting met de gewenste systolische druk............................................... 8
De monitor gebruiken ....................................................................... 9
De batterijen plaatsen / vervangen ............................................................... 9
De luchtslang aansluiten .............................................................................. 10
De netstroomadapter aansluiten ................................................................. 10
De juiste manchetgrootte selecteren .......................................................... 11
De armmanchet aanbrengen ...................................................................... 11
Nauwkeurige metingen uitvoeren ............................................................... 13
Meting ......................................................................................................... 13
Na meting ....................................................................................................... 13
Metingen ......................................................................................... 14
Normale meting ............................................................................................. 14
Meting met de gewenste systolische druk ................................................. 15
Een nauwkeurige meting uitvoeren ............................................................ 15
De gegevens uit het geheugen opvragen ....................................... 16
Waarvoor dient de IHB/AFib-indicator? .......................................... 17
Wat is de AFib? .............................................................................. 17％IHB/AFib ...................................................................................... 17
Drukbalkindicator ............................................................................ 18
Indicator WHO-classificatie ............................................................. 18
Meer over bloeddruk ....................................................................... 18
Wat is bloeddruk? ......................................................................................... 18
Wat is hoge bloeddruk en hoe wordt deze onder controle gehouden? . 19
Waarom thuis bloeddruk meten?................................................................ 19
Bloeddrukclassificatie WHO ........................................................................ 19
Bloeddrukvariaties......................................................................................... 19
Probleemoplossing ......................................................................... 20
Onderhoud ...................................................................................... 21
Technische gegevens ..................................................................... 21
Nederlands 1

Beste klant
Gefeliciteerd met uw aanschaf van een hypermoderne A&D-bloeddrukmonitor.
Dit apparaat is ontworpen voor gebruiksgemak en nauwkeurigheid en zal uw
dagelijkse bloeddrukmetingen vergemakkelijken.
We bevelen u aan deze handleiding zorgvuldig door te lezen voordat
u het apparaat voor het eerst gebruikt.

Inleidende opmerkingenDit apparaat voldoet aan Europese Richtlijn 93/42/EEG betreffende
medischehulpmiddelen.
Dit
wordt
aangegevenmetdeconformiteitsmarkering. (0123: het referentienummer voor debetrokken aangemelde instantie)
Het apparaat is ontworpen voor gebruik bij volwassenen, niet bij baby's ofpeuters.
Gebruiksomgeving. Het apparaat is bedoeld voor gebruik door uzelf voor
medische verzorging in de thuissituatie.
Dit apparaat is ontworpen voor het meten van bloeddruk en hartslag vanpersonen voor diagnosedoeleinden.

Voorzorgsmaatregelen
Voor de constructie van dit apparaat zijn precisieonderdelen gebruikt.
Extreme temperatuur, luchtvochtigheid, direct zonlicht, schokken of stof
dienen te worden vermeden.
Reinig het apparaat en de manchet met een droge, zachte doek of een
doek die is bevochtigd met water en een neutraal schoonmaakmiddel.
Gebruik in geen geval alcohol, benzeen, thinner of andere agressieve
chemische middelen om het apparaat en de manchet te reinigen.
Vouw de manchet niet te strak op en bewaar de slang niet gedurende
langere perioden in strak opgewikkelde vorm, aangezien dit delevensduur van de onderdelen kan verkorten.
Pas op dat baby's of peuters zichzelf niet per ongeluk wurgen met de slang
en kabel.
Draai de luchtslang niet tijdens het meten. Dit zou letsel kunnenveroorzaken als gevolg van continue druk van de manchet.
Het apparaat en de manchet zijn niet waterbestendig. Voorkom dat het
apparaat en de manchet in contact komen met regen, zweet en water.
Metingen kunnen verkeerde waarden aangeven als het apparaat wordt
gebruikt in de nabijheid van televisies, magnetrons, mobiele telefoons,röntgenapparaten of andere apparaten met een sterk elektrisch veld.
Draadloze communicatieapparaten zoals thuisnetwerkapparaten, mobiele
telefoons, draadloze telefoons en de basisstations daarvan en walkietalkies
kunnen van invloed zijn op de bloeddrukmonitor.
Hanteer daarom een minimumafstand van 30 cm van dergelijke apparaten.
Wanneer u het apparaat hergebruikt, dient u eerst te controleren of het
schoon is.
Gebruikte apparatuur, onderdelen en batterijen mogen niet als normaal
Nederlands 2

huishoudelijk afval worden behandeld en moeten worden afgevoerd
volgens de toepasselijke plaatselijke regels.
Wanneer u de netstroomadapter gebruikt, dient u ervoor te zorgen dat deadapter snel uit het stopcontact kan worden gehaald indien nodig.
Breng geen wijzigingen aan in het apparaat. Dit kan ongevallen of schadeaan het apparaat veroorzaken.
Om de bloeddruk te meten, moet de arm hard genoeg worden afgekneld
door de manchet zodat de bloedstroom door de slagader tijdelijk wordt
gestopt. Dit kan pijn, gevoelloosheid of een tijdelijke rode verkleuring vande arm opleveren. Dit doet zich met name voor wanneer de meting
meerdere keren in korte tijd wordt herhaald. Eventuele pijn, gevoelloosheidof rode verkleuringen verdwijnen na verloop van tijd.
Het te vaak meten van de bloeddruk kan kwalijk zijn doordat debloedcirculatie wordt verstoord. Controleer dat de werking van het apparaat
niet resulteert in langdurige verstoring van de bloedcirculatie wanneer u het
apparaat herhaaldelijk gebruikt.
Als u een borstamputatie hebt ondergaan, dient u een arts te raadplegenvoordat u het instrument gebruikt.
Zorg ervoor dat kinderen het apparaat niet zonder toezicht kunnen
gebruiken en gebruik het apparaat niet binnen bereik van baby's of peuters.
Dit kan ongevallen of schade veroorzaken.
Het apparaat bevat kleine onderdelen die een risico op verstikking
opleveren indien ze door baby's of peuters worden ingeslikt.
Koppel de netstroomadapter los wanneer deze niet in gebruik is tijdens
het meten.
Het gebruik van accessoires die niet in deze handleiding worden vermeld,
kan een veiligheidsrisico opleveren.
Indien de batterij kortsluiting maakt, kan deze heet worden, met mogelijk
brandwonden tot gevolg.
Laat het apparaat op kamertemperatuur komen voordat u het gebruikt(ongeveer een uur).
Er zijn geen klinische tests uitgevoerd bij pasgeborenen en zwangere
vrouwen. Gebruik het niet bij pasgeborenen of zwangere vrouwen.
Zorg ervoor dat u de batterijen, de DC-aansluiting en de patiënt niet
tegelijkertijd aanraakt. Dit zou een elektrische schok kunnen veroorzaken.
Pomp de manchet niet op voordat u deze rond de bovenarm hebt gewikkeld.

Contra-indicaties
Hierna volgen voorzorgsmaatregelen voor het juiste gebruik van het apparaat.
Breng de manchet niet aan rond een arm waar ook andere medischeelektronica zijn aangesloten. Het instrument werkt dan mogelijk niet goed.
Mensen die ernstige circulatieproblemen hebben in de arm, moeten eerst
een arts raadplegen voordat ze het apparaat gebruiken. Dit om medische
problemen te voorkomen.
Voer geen zelfdiagnose uit aan de hand van de meetwaarden en begin
Nederlands 3

niet zelf een behandeling. Raadpleeg altijd eerst uw arts voor beoordelingen behandeling.
Breng de manchet niet aan rond een arm met een nog niet genezen
wond.
Breng de manchet niet aan rond een arm die voor een infuus of
bloedtransfusie wordt gebruikt. Dit kan letsel of ongevallen veroorzaken.
Gebruik het apparaat niet op een plaats waar brandbare gassen
aanwezig zijn, zoals narcosegassen. Dit zou een explosie kunnenveroorzaken.
Gebruik het apparaat niet in omgevingen met een hoge
zuurstofconcentratie, zoals een zuurstofkamer met hoge druk of een
zuurstoftent. Dit kan brand of een explosie veroorzaken.

Nederlands 4

Identificatie onderdelen
Display

Connector
luchtaansluiting

Connector
netstroomadapter

DC-ingang
START -knop

Aansluitpunt lucht

Luchtslang

Juiste maatbereik

Slagadermarkering
Indexmarkering
Armmanchet
Batterijvak

Batterijklep

1,5 V-batterijen(R6P, LR6 of AA)

Display

Gemiddeld
GEHEUGENIHB/AFib-symbool
Systolische
druk%IHB/AFib
Symbool onjuist
aangebrachte manchet

Diastolische
druk

Symbool bewegingsfout
Hartmarkering

HartslagIndicator WHO-classificatie
en drukbalkindicator
Batterij-indicator
Nederlands 5

Symbolen
Symbolen die op de behuizing van het apparaat zijn gedrukt
Symbolen
SYS
DIA
PUL

IP

Functie / betekenis
Stand-by en Apparaat aanzetten.
Systolische bloeddruk in mmHg
Diastolische bloeddruk in mmHg
Hartslag per minuut
Installatie-instructies batterij
Gelijkstroom (DC)Type BF: apparaat, manchet en slangen zijn ontworpen om
speciale bescherming te bieden tegen elektrische schokken.
Label EG-richtlijn medische apparaten
EU-vertegenwoordiger
Fabrikant
Productiedatum
Internationaal beveiligingssymbool
WEEE-label

SN

Serienummer
Zie de instructiehandleiding/het instructieboekje
Polariteit van DC-aansluiting
Droog houden

Symbolen die worden weergegeven op de display
Symbolen

Functie / betekenis
Aanbevolen actie
Wordt weergegeven wanneer
een meting wordt uitgevoerd.
De meting wordt uitgevoerd. Blijf
Het knippert wanneer dezo stil mogelijk zitten.
hartslag wordt waargenomen.
IHB/AFib-symbool
Wordt weergegeven wanneer
een onregelmatige hartslag
wordt waargenomen.
Het kan gaan branden wanneer
een zeer lichte vibratie, zoals
een rilling of siddering, wordt
waargenomen.
Er kan een onjuiste meetwaarde
Wordt weergegeven wanneer
worden weergegeven. Voer de
een beweging van lichaam of
meting opnieuw uit.
arm wordt waargenomen.
Blijf stil zitten tijdens het meten.
Nederlands 6

Symbolen
Symbolen

Functie / betekenis

Aanbevolen actie
Er kan een onjuiste
meetwaarde worden
Wordt tijdens meting weergegevenweergegeven. Breng de
als de manchet te los zit
manchet correct aan envoer de meting opnieuw uit.
Gedetecteerd percentage IHB/AFibin geheugen
Aantal gedetecteerde
IHB/AFib-voorvallen in
geheugen%IHB/AFib =x 100 [%]

Totaal aantal

Eerdere metingen die in
GEHEUGEN zijn opgeslagen.
Gemiddelde waarden
BATTERIJ VOL
De batterijstroomindicator tijdens
meting.
BATTERIJ BIJNA LEEG
De batterij is bijna leeg als dit
knippert.
Onstabiele bloeddruk als gevolg
van beweging tijdens meting.

Vervang alle batterijen door
nieuwe exemplaren
wanneer dit symbool
knippert.
Voer de meting opnieuw uit.
Blijf helemaal stil zitten
tijdens het meten.

De systolische en diastolische
waarden liggen binnen 10 mmHg
van elkaar.
De drukwaarde is niet toegenomen
Breng de manchet correct
tijdens het opblazen.
aan en voer de meting
De manchet is niet correct
opnieuw uit.
aangebracht.
FOUT HARTSLAGWEERGAVE
De hartslag is niet goedgedetecteerd.
Verwijder de batterijen,
druk op de knop START en
plaats de batterijen terug.
Interne fout bloeddrukmonitor
Neem contact op met deverkoper als de fout nog
steeds wordt weergegeven.
Nederlands 7

Gebruiksmodus
1. Normale meting
Druk op de knop START. De bloeddruk wordt gemeten en de gegevens
worden in het geheugen opgeslagen. Dit apparaat kan de laatste 60 metingen
in het geheugen opslaan.

2. De gegevens opvragenAls niets wordt weergegeven, drukt
u opnieuw op de knop START en houdt u deze
ingedrukt.

Houd de knop
ingedrukt terwijl het
apparaat
START in stand-by

staat

Laat de knop los wanneer de gemiddeldewaarden worden weergegeven.

Het gegevensnummer en de opgeslagengegevens worden automatisch op volgorde
weergegeven vanaf de laatste meting.

Zie "De gegevens uit het geheugen
opvragen" voor meer informatie over het
opvragen van gegevens.

3. Alle gegevens in het geheugen wissen
Terwijl u het apparaat uitzet, houdt u de knop
START ingedrukt tot "" wordt weergegeven.
Selecteer "" om gegevens te wissen.
De gegevens zijn gewist als het symbool
knippert.
Het apparaat wordt automatisch uitgezet.

4. Meting met de gewenste systolische druk
Zie pagina 15 voor metingen met de gewenste systolische druk.
Nederlands 8START

De monitor gebruiken
De batterijen plaatsen / vervangen
1. Verwijder de batterijklep.
2. Verwijder de gebruikte batterijen enplaats nieuwe batterijen in het
batterijvak, zoals hier weergegeven.
Zorg ervoor dat de polariteit (+ en -)correct is.
Gebruik uitsluitend R6P-, LR6- of
AA-batterijen.

Stap 1
Stap 3

Stap 1
Stap 2

3. Plaats de batterijklep terug.

LET OP
Plaats de batterijen zoals weergegeven in het batterijvak. Als ze niet goed
worden geplaatst, werkt het apparaat niet.
Als(BATTERIJ BIJNA LEEG) knippert op de display, dient u allebatterijen door nieuwe te vervangen. Mix nooit oude en nieuwe batterijen.
Dit kan de levensduur van de batterijen verkorten. Daarnaast kan het
ervoor zorgen dat het apparaat defect raakt.
Vervang de batterijen twee seconden of langer nadat u het apparaat hebt
uitgezet.
Als het symbool(BATTERIJ BIJNA LEEG) zelfs na vervanging van
de batterijen wordt weergegeven, dient u een bloeddrukmeting uit tevoeren. Het is mogelijk dat het apparaat de nieuwe batterijen dan wel
herkent.
Het symbool (BATTERIJ BIJNA LEEG) wordt niet weergegeven als debatterijen helemaal leeg zijn.
De levensduur van de batterijen varieert al naargelang deomgevingstemperatuur en kan korter zijn bij lage temperaturen. Vier
nieuwe LR6-batterijen gaan doorgaans ongeveer een jaar mee wanneer
u het apparaat tweemaal daags voor metingen gebruikt.
Gebruik alleen de aangegeven batterijen. De batterijen die met het
apparaat worden meegeleverd, zijn bedoeld voor het testen van demonitor en hebben mogelijk een beperkte levensduur.
Verwijder de batterijen als u het apparaat langere tijd niet gebruikt.
De batterijen kunnen gaan lekken en een defect veroorzaken.

Nederlands 9

De monitor gebruiken
De luchtslang aansluiten

Aansluitpunt lucht

Plaats de connector van deluchtslang stevig in het
aansluitpunt voor lucht.
Connector
luchtaansluiting

De netstroomadapter aansluiten
Plaats de connector voor denetstroomadapter in deDC-ingang.
Sluit de netstroomadapter
vervolgens aan op een
stopcontact.

DC-ingang
Connector netstroomadapter

Gebruik de gespecificeerde
netstroomadapter.
(Zie pagina 22.)
Wanneer u de netstroomadapter loskoppelt van het stopcontact, doetu dit door de adapter zelf vast te grijpen en deze uit het stopcontact tetrekken.
Wanneer u de netstroomadapter loskoppelt van de bloeddrukmonitor,
doet u dit door de connector vast te grijpen en deze uit de monitor tetrekken.

Nederlands 10

De monitor gebruiken
De juiste manchetgrootte selecteren
Voor het verkrijgen van een nauwkeurige meting, is het belangrijk dat u dejuiste manchetgrootte gebruikt. Als de manchet niet de juiste maat heeft,kan de meting een onjuiste bloeddrukwaarde opleveren.
De armomvang is op elke manchet gedrukt.
De indexen het juiste maatbereik op de manchet laten u zien of u dejuiste manchet gebruikt. (Zie "Symbolen die op de manchet zijn gedrukt" op
de volgende pagina)
Als de index
buiten het maatbereik wijst, dient u contact op te nemenmet uw plaatselijke verkoper om een vervangende manchet te kopen.
De armmanchet is een verbruiksitem. Koop een nieuw exemplaar als deoude versleten raakt.
Armomvang
Aanbevolen manchetgrootte
Catalogusnummer
31 cm tot 45 cm
Grote manchet volwassenen
CUF-F-LA
22 cm tot 42 cm
Manchet voor breed spectrum
CUF-I
22 cm tot 32 cm
Manchet volwassenen
CUF-F-A
Armomvang: de omtrek ter hoogte van de biceps.
Opmerking: de UA-767S-W is niet ontworpen voor gebruik van een
kleine manchet.

De armmanchet aanbrengen
1. Wikkel de manchet rond debovenarm, ongeveer 1 - 2 cm boven
de binnenzijde van de elleboog,
zoals in de illustratie.
Plaats de manchet direct tegen dehuid. Kleding kan een zwakkere
hartslagmeting veroorzaken en kan
zo een meetfout tot gevolg hebben.

Rol uw mouwen niet te strak op.
Manchet1 - 2 cm

Luchtslang
Slagadermarkering
Maak deklittenbandsluiting
stevig vast.

2. Het afklemmen van de bovenarm
door het te strak oprollen van een
mouw kan voor een onnauwkeurige
meting zorgen.
3. Ga na of de index
binnen het
juiste maatbereik valt.

Nederlands 11

Index

en het juiste maatbereikDe monitor gebruiken
Symbolen die op de manchet zijn gedrukt
Symbolen

Functie / betekenis

Aanbevolen actie
Plaats de -markering op
de slagader van debovenarm of in één lijn
met de ringvinger aan debinnenzijde van de arm.

Slagadermarkering
▲
REF

Index
Catalogusnummer
Het juiste maatbereik voor demanchet voor volwassenen.
Het is gedrukt op de manchet voor
volwassenen.
Gebruik de grote manchet
voor volwassenen in plaats
Groter dan het bereik dat is gedrukt
van de manchet voor
op de manchet voor volwassenen en
volwassenen of dede manchet voor breed spectrum.
manchet voor breed
spectrum.
Het juiste maatbereik voor de manchet
voor breed spectrum. Het is gedrukt op
de manchet voor breed spectrum.
Het juiste maatbereik voor de grotemanchet voor volwassenen. Het isgedrukt op de grote manchet voor
volwassenen.
Kleiner dan het bereik dat is gedrukt
op de manchet voor volwassenen ende manchet voor breed spectrum.
Gebruik de manchet voor
Kleiner dan het bereik dat is gedrukt
volwassenen in plaats van
op de grote manchet voor
de grote manchet voor
volwassenen.
volwassenen.
Partijnummer

Grote manchet volwassenen

Juiste maatbereik

A

L

Manchet voor breed spectrum

S
Manchet volwassenen

S

Nederlands 12

W

L

A

L

De monitor gebruiken
Nauwkeurige metingen uitvoeren
Ga als volgt te werk voor de meest nauwkeurige bloeddrukmeting:
Ga comfortabel op een stoel zitten. Laat uw arm op de tafel rusten. Zorg
ervoor dat u uw benen niet kruist. Houd uw voeten plat op de vloer en houduw rug recht.
Ontspan gedurende vijf tot tien minuten voordat u gaat meten.
Plaats het midden van de manchet ter hoogte van uw hart.
Blijf stil zitten en praat niet tijdens het meten.
Voer de meting niet uit als u zich net fysiek hebt ingespannen of een bad
hebt genomen. Rust ongeveer twintig tot dertig minuten voordat u demeting uitvoert.
Probeer uw bloeddruk iedere dag rond dezelfde tijd uit te voeren.

MetingHet is normaal als de manchet erg knelt tijdens het meten. (Maak u niet
ongerust).

Na meting
Druk na meting op de knop START om het apparaat uit te zetten.
Verwijder de manchet en registreer uw gegevens.
Opmerking: het apparaat heeft een automatische uitschakelfunctie, waardoor het
apparaat ongeveer een minuut na de meting wordt uitgezet.
Gebruik een tussenpauze van minstens drie minuten voor metingenbij dezelfde persoon.

Nederlands 13

Metingen
Lees voordat u tot meten overgaat "Een nauwkeurige meting uitvoeren" op devolgende pagina.

Normale meting
1.

Plaats de manchet rond de arm (bij voorkeur de
Ter hoogte
linkerarm).
van het hart
Zit stil tijdens het meten.
2. Druk op de knop START.
Druk op START
Alle displayonderdelen worden
weergegeven.
Nul (0) wordt weergegeven en knippert een
Alledisplayonderdelen
korte tijd.
in beeld
De weergave op de display verandert, zoals tezien op de afbeelding rechts, op het moment
dat de meting begint. De manchet wordt
opgeblazen. Het is normaal als de manchet
Nulweergave
erg knelt. Tijdens het opblazen wordt links op
Start
opblazende display een drukbalkindicator
weergegeven.
Opmerking:als u het opblazen op enig
moment wilt stoppen, drukt uopnieuw op de knop START.
Bezig met
3. Zodra de manchet volledig is opgeblazen, begint
opvoeren druk
het leeglopen automatisch. Het hartsymbool
gaat knipperen, wat aangeeft dat er wordt
gemeten. Zodra de hartslag wordt gedetecteerd,knippert het symbool bij elke hartslag.
Opmerking: als er geen passende druk wordt bereikt,
begint het apparaat automatischopnieuw met opblazen.
Lees "Meting met de gewenste
systolische druk" op de volgende pagina
De metingom opnieuw opblazen te voorkomen.
wordt
uitgevoerd
4. Zodra de meting voltooid is, worden desystolische en diastolische drukwaarden ende hartslag weergegeven.
Systolische druk
De manchet laat de resterende lucht
Diastolische druk
ontsnappen en loopt volledig leeg.
WHO-classificatie
5. Druk op de knop START om het apparaat uit
te zetten.Na één minuut inactiviteit wordt het
Hartslagapparaat automatisch uitgezet.
Opmerking: zorg voor een tussenpauze vanResterende lucht
minstens drie minuten voor metingen
wordt automatischbij dezelfde persoon.
uitgestoten
Nederlands 14

Metingen
Meting met de gewenste systolische druk
De UA-767S-W is ontworpen om de hartslag te detecteren en demanchet automatisch tot een systolisch drukniveau op te blazen.
Gebruik de volgende methode wanneer er herhaaldelijk opnieuw wordt
opgeblazen of wanneer de resultaten niet worden weergegeven, zelfs als
de druk tot 20 mmHg of minder is gedaald.
1.

Plaats de manchet ter hoogte van het hartrond de arm (bij voorkeur de linkerarm).

2.

Druk op de knop START.

3.

Zodra de nul knippert, dient u deknop START ingedrukt te houden tot
een getal wordt weergegeven dat
ongeveer 30 tot 40 mmHg hoger isdan uw verwachte systolische druk.

4.

Ter hoogte
van het hart
Druk op STARTde knop

Zodra het gewenste getal is bereikt, laatu de knop START los om de meting testarten. Ga door met het meten van uw
bloeddruk zoals beschreven op devorige pagina.

Een nauwkeurige meting uitvoeren

Laat de knop los bij
de gewenste
systolische druk

Houd de knop
ingedrukt

START

Zie de vorige pagina voor
meting

Ga in een comfortabele houding zitten Plaats uw arm op een tafel met dehandpalm omhoog en de manchet ter hoogte van uw hart.
Ontspan gedurende vijf tot tien minuten voordat u gaat meten. Als
u opgewonden of depressief bent als gevolg van emotionele stress, zal
de meting deze stress weerspiegelen in de vorm van een hogere (oflagere) bloeddrukwaarde dan normaal en zal de hartslag meestal snellerzijn dan normaal.
De bloeddruk van een persoon varieert constant, afhankelijk van wat
u doet en wat u hebt gegeten. Wat u drinkt kan een zeer sterk en snel
effect hebben op uw bloeddruk.
Dit apparaat baseert zijn metingen op de hartslag. Als u een erg zwakke
of onregelmatige hartslag hebt, kan het apparaat problemen hebben bij
het bepalen van uw bloeddruk.
Nederlands 15

Als het apparaat een toestand aantreft die abnormaal is, stopt het met
meten en geeft het een foutsymbool weer. Zie pagina 7 voor debeschrijving van symbolen.
Dit apparaat is bedoeld voor gebruik door volwassenen. Raadpleeg uw
arts voordat u dit instrument op een kind gebruikt. Kinderen hebben
toezicht nodig bij het gebruik van dit apparaat.
De prestatie van de automatische bloeddrukmonitor kan wordenbeïnvloed door buitensporige temperaturen, luchtvochtigheid of hoogte.

De gegevens uit het geheugen opvragen
Opmerking: dit apparaat slaat de laatste 60 metingen op in zijn geheugen.
Knop START ingedrukt
1. Als er niets wordt weergegeven,houdenGemiddelde systolische
houdt u de knop START ingedrukt
waardeom de opgeslagen gegevens op te
Gemiddelde diastolischevragen.
waarde2.

Laat de knop los wanneer degemiddelde waarden worden
weergegeven.

3.

Het gegevensnummer en deopgeslagen gegevens worden
automatisch op volgorde
weergegeven vanaf de laatste meting.

4.

De display wordt automatischuitgeschakeld nadat alle gegevens
zijn weergegeven.

Gemiddelde hartslag

Gemiddelde vanalle gegevens

SystolischDiastolischHartslagMeest recente gegevens

Opmerking: als u tijdens het opvragenvan gegevens op de knop
START drukt, wordt het
apparaat uitgeschakeld.
SystolischDiastolischHartslagLaatste gegevens (oudste)

Nederlands 16

Waarvoor dient de IHB/AFib-indicator?
Zodra de monitor tijdens de metingen een onregelmatige hartslag waarneemt,
wordt de IHB/AFib-indicator op de display weergegeven naast de
meetwaarden.
Opmerking: we bevelen aan dat u contact met uw huisarts opneemt als
u dezeIHB/AFib-indicator regelmatig ziet.

Wat is de AFib?
Het hart trekt samen als gevolg van elektrische signalen in het hart en stuurt
bloed door het lichaam. Atriumfibrillatie (AFib) doet zich voor wanneer het
elektrische signaal in het atrium verward raakt, wat leidt tot storingen in dehartslaginterval. AFib kan ervoor zorgen dat bloed in het hart stagneert, wat
makkelijk kan leiden tot bloedproppen, een oorzaak voor beroertes en
hartaanvallen.

％IHB/AFib%IHB/AFib wordt weergegeven als frequentie van de waargenomen IHB/AFib.
IHB/AFib kan niet alleen geluiden waarnemen, zoals fysieke bewegingen,maar ook een onregelmatige hartslag. We raden u dan ook aan contact
met uw arts op te nemen als %IHB/AFib een hoge waarde aangeeft.
%IHB/AFib =Aantal gedetecteerde
IHB/AFib-voorvallen inx 100 [%]

Totaal aantal

Weergave van %IHB/AFib: %IHB/AFib wordt weergegeven wanneer
gemiddelde waarden worden weergegeven.
(Zie "2. De gegevens opvragen" in "Gebruiksmodus")%IHB/AFib wordt niet weergegeven wanneer
het geheugennummer zes of lager is.
Weergave gemiddelde waarden

Niveau 0%IHB/AFib=0

Niveau 1%IHB/AFib=1 - 9

Niveau 2%IHB/AFib=10 - 24

Niet
weergegeven

Nederlands 17

Niveau 3%IHB/AFib=25 - 100

Drukbalkindicator
De indicator bewaakt de drukontwikkeling tijdens metingen.
Bezig met opvoeren druk Bezig met leeglopen

Bezig met opblazen

Bezig met meten
Opblazen is voltooid

Indicator WHO-classificatie
Elk segment van de balkindicator komt
overeen met de bloeddrukclassificatie vande WHO die op de volgende pagina wordt
beschreven.
Indicator WHO-classificatie
Ernstige hoge bloeddruk
Gemiddelde hoge bloeddruk
Milde hoge bloeddruk
Hoog normaal
NormaalOptimaal: De indicator geeft een segment weer op
basis van de huidige gegevens,corresponderend met deWHO-classificatie.

Voorbeeld:

Gemiddelde hoge bloeddruk

Milde hoge bloeddruk

Hoog normaal

Meer over bloeddruk
Wat is bloeddruk?
Bloeddruk is de kracht die door het bloed wordt uitgeoefend tegen dewanden van de slagaders. Systolische druk doet zich voor wanneer het hartsamentrekt. Diastolische druk doet zich voor wanneer het hart uitzet.
Bloeddruk wordt gemeten in millimeter kwik (mmHg). Uw natuurlijke
bloeddruk wordt vertegenwoordigd door de fundamentele druk, die 'sochtends wordt gemeten terwijl u nog in ruste bent en voordat u ontbijt.
Nederlands 18

Wat is hoge bloeddruk en hoe wordt deze onder controle
gehouden?

Hoge bloeddruk is een abnormaal hoge arteriële bloeddruk die, indien
onbehandeld, tot een groot aantal gezondheidsproblemen kan leiden,
waaronder beroertes en hartaanvallen. Hoge bloeddruk kan onder controle
worden gehouden door het veranderen van uw levensstijl, het vermijden
van stress en met medicatie onder dokterstoezicht.
U kunt het volgende doen om hoge bloeddruk te voorkomen of onder
controle te houden:
Niet roken
Regelmatig bewegen
Minder zout en vet in uw
Regelmatig een medische controle
voedsel gebruiken
ondergaan
Een gezond gewicht
handhaven

Waarom thuis bloeddruk meten?

Als de bloeddruk in het ziekenhuis of bij een huisarts wordt gemeten, kan dit
spanning veroorzaken, wat tot een verhoogde waarde kan leiden, 25 tot 30
mmHg hoger dan wanneer thuis gemeten. Thuismeting reduceert deeffecten van externe invloeden op de bloeddrukwaarden, geeft een
aanvulling op de metingen van de arts en leidt tot een nauwkeuriger enmeer complete bloeddrukgeschiedenis.

Bloeddrukclassificatie WHO

Bloeddrukvariaties

Diastolische bloeddruk

De Wereldgezondheidsorganisatie Naslagmateriaal: Journal of Hypertension 1999, Vol 17, nr. 2
Hoge bloeddruk type 3 (ernstig)(WHO) heeft normen opgesteldvoor het beoordelen van bloeddruk,
Hoge bloeddruk type 2 (gemiddeld)zonder onderscheid te maken naar
Hoge bloeddruk type 1 (mild)
leeftijd. Deze normen worden
weergegeven in de grafiek.
Hoog-normaal
NormaalOptimaal

Bloeddruk

De bloeddruk van een persoon kan
Systolische bloeddruk
flink variëren, zowel in de loop vande dag als tijdens het seizoen. Er kan een verschil van 30 tot 50 mmHg
zijn als gevolg van diverse omstandigheden in de loop van de dag. Bij
personen met hoge bloeddruk zijn deze variaties zelfs nog groter.
Normaal gesproken stijgt de bloeddruk terwijl u werkt of aan sport doeten is deze op zijn laagste punt wanneer u slaapt. Maak u daarom niet alte ongerust over de resultaten van één
Normale fluctuatie binnen een dag
enkele meting.
(om de vijf minuten gemeten)
Doe uw metingen elke dag op dezelfde
Systolischtijd en gebruik daarvoor de procedure diein deze handleiding wordt beschrevenom uw normale bloeddruk te achterhalen.
Regelmatige metingen geven een meer
volledige bloeddrukgeschiedenis. Zorg
DiastolischSlaapervoor dat u ook de datum en tijd
Tijd
opschrijft wanneer u uw bloeddruk
noteert. Raadpleeg uw arts om uw
bloeddrukgegevens te interpreteren.
Nederlands 19

Probleemoplossing
Probleem

Mogelijke oorzaak

Aanbevolen actie
Vervang alle batterijen door
De batterijen zijn leeg.
Er wordt niets
nieuwe exemplaren.
op de display
Plaats de batterijen opnieuw enweergegeven,
De contactpunten van de zorg ervoor dat de positieve
terwijl het
batterijen zitten niet op
contactpunten op de posities
apparaat aan
de juiste positie.
zitten die in het batterijvak zijn
staat.
aangeduid.
Het batterijvoltage is telaag.
Het symbool
De manchet
Vervang alle batterijen door
(BATTERIJ BIJNA
wordt niet
nieuwe exemplaren.
LEEG) knippert. Als deopgeblazen.
batterijen helemaal leegzijn, wordt het symbool
niet weergegeven.
De manchet is niet
Breng de manchet correct aan.
correct aangebracht.
U hebt uw arm of lichaam Zorg ervoor dat u erg stil blijftbewogen tijdens het
zitten en geen geluid maakt
meten.
tijdens metingen.
Het instrument
Zorg ervoor dat u comfortabel envoert geenstil zit.
meting uit.
De positie van de
De waarden zijn manchet is niet correct. Plaats uw arm op een tafel met dehandpalm omhoog en de manchet
te hoog of teter hoogte van uw hart.
laag.
Als u een erg zwakke of
onregelmatige hartslag hebt, kan
het apparaat problemen hebben
bij het bepalen van uw
bloeddruk.
De waarde wijkt af vande waarde die bij een
Zie "Waarom thuis bloeddruk
bezoek aan een
meten?".
ziekenhuis of huisarts is
Overiggemeten.
Verwijder de batterijen. Plaatsze op de goede manier terug envoer de meting opnieuw uit.
Opmerking: neem contact op met de verkoper als u het probleem niet kunt
oplossen met de hierboven beschreven acties. Probeer dit
product niet te openen of repareren. Als u dit toch doet, wordt
uw garantie ongeldig.
Nederlands 20

OnderhoudOpen het apparaat niet. Er worden in dit apparaat delicate onderdelen eneen ingewikkelde luchteenheid gebruikt die beschadigd zouden kunnen
raken. Als u het probleem niet kunt oplossen aan de hand van deprobleemoplossingsinstructies, dient u contact op te nemen met degeautoriseerde verkoper in uw gebied of met onze afdeling klantenservice.
De klantenservice van A&D voorziet geautoriseerde verkopers vantechnische informatie, reserveonderdelen en reserve-eenheden.
Het apparaat is ontworpen en gefabriceerd voor een lange levensduur.
Wij bevelen echter over het algemeen aan het apparaat elke 2 jaar telaten inspecteren om te controleren dat dit nog goed werkt en nauwkeurig
is. Neem voor onderhoud contact op met de geautoriseerde verkoper in
uw gebied of met A&D.

Technische gegevens
Type
Meetmethode
Meetbereik

UA-767S-W
Oscillometrische meting
Druk: 0 - 299 mmHg
Systolische druk: 60 - 279 mmHg
Diastolische druk: 40 - 200 mmHg
Hartslag: 40 - 180 slagen / minuutDruk:± 3 mmHg
Meetnauwkeurigheid
Hartslag: ± 5 %4 x 1,5 V-batterijen (R6P, LR6 of AA) of
Stroomvoorziening
netstroomadapter (TB-233C) (niet inbegrepen)
Ongeveer 700 keer LR6 (alkalinebatterijen)
Aantal metingen
Ongeveer 200 keer R6P (mangaanbatterijen)
Bij een drukwaarde van 180 mmHg,
kamertemperatuur 23 °C.
Classificatie
ME-apparatuur met interne voeding (voeding via
batterijen) / Klasse II (voeding via adapter)
Modus continu bedrijf
Volgens ISO81060-2：2013
Klinische test
Tijdens de klinische validatiestudie, werd K5 op 85
proefpersonen gebruikt om de diastolische
bloeddruk te bepalen.
IEC 60601-1-2: 2014
EMD
Laatste 60 metingen
GeheugenGebruiksomstandigheden + 10 tot + 40 °C / relatieve luchtvochtigheid 15 tot
85 % / 800 tot 1.060 hPa
Omstandigheden voor
-20 tot + 60 °C / relatieve luchtvochtigheid 10 tot
95 % / 700 tot 1.060 hPatransport / opslag
Ongeveer 140 [B] x 60 [H] x 105 [D] mm
Afmetingen
Nederlands 21

Gewicht
Beschermingsgraad
Aangebracht onderdeel
Nuttige levensduur

Ongeveer 245 g, exclusief batterijen
Apparaat: IP20
Manchet Type BF
Apparaat: 5 jaar (indien zes keer per dag
gebruikt)Manchet: 2 jaar (indien zes keer per dag
gebruikt)
Netstroomadapter: 5 jaar (indien zes keer per dag
gebruikt)

Accessoire netstroomadapter
De adapter wordt gebruikt om het apparaat thuis op een voedingsbron aan
te sluiten. Neem voor aankoop contact op met uw plaatselijke A&D-verkoper.
De netstroomadapter moet periodiek worden geïnspecteerd of vervangen.
TB-233C
Neem voor aankoop contact op met uw
plaatselijke A&D-verkoper.
De netstroomadapter moet periodiek wordengeïnspecteerd of vervangen.
Symbolen die op de netstroomadapter zijn gedrukt
Symbolen
Functie / betekenis
Uitsluitend voor gebruik binnenshuis
Klasse II-apparaat
Thermische zekering
Zekering
Apparaatlabel EG-richtlijn
Apparaatlabel EAC-certificatie
Polariteit connector netstroomadapter
Accessoires worden afzonderlijk verkocht
Manchet Catalogusnummer
Manchetgrootte
Armomvang
CUF-F-LA
Grote manchet volwassenen 31 cm tot 45 cm
CUF-I
Manchet voor breed spectrum 22 cm tot 42 cm
CUF-F-A
Manchet volwassenen
22 cm tot 32 cm
Armomvang: de omtrek ter hoogte van de biceps.
Stekker (typewandcontact)
TB-233C
Type C
Opmerking: specificaties kunnen zonder voorafgaande kennisgeving worden
gewijzigd.
De IP-classificatie is de mate van bescherming die door
behuizingen wordt geboden conform IEC 60529. Dit apparaat isbeschermd tegen vaste vreemde voorwerpen met een diametervan 12 mm of groter, zoals vingers. Dit apparaat is nietbeschermd tegen water.
Netstroomadapter

Catalogusnummer

Nederlands 22

IndholdKære kunde ...................................................................................... 2
Indledende bemærkninger ................................................................ 2
Sikkerhedsforholdsregler .................................................................. 2
Oversigt over dele ............................................................................. 5
Symboler ........................................................................................... 6
Driftstilstand ...................................................................................... 8
1. Normal måling ......................................................................... 8
2. Hentning af data ...................................................................... 8
3. Sletning af alle data, der er lagret i hukommelsen .................. 8
4. Måling med det ønskede systoliske tryk.................................. 8Sådan bruges måleren ..................................................................... 9Isætning/udskiftning af batterier .................................................. 9
Tilslutning af luftslangen ............................................................ 10
Tilslutning af AC-adapteren ....................................................... 10
Valg af den korrekt manchetstørrelse ....................................... 11Påsætning af arm-manchetten .................................................. 11Sådan foretager du nøjagtige målinger ..................................... 13Måling........................................................................................ 13
Efter måling ............................................................................... 13Målinger .......................................................................................... 14
Normal måling ........................................................................... 14Måling med det ønskede systoliske tryk.................................... 15Bemærkninger om nøjagtig måling ........................................... 15
Hentning af hukommelsesdata ....................................................... 16
Hvad er indikatoren for IHB/AFib? .................................................... 17
Hvad er AFib? ................................................................................. 17％IHB/AFib ...................................................................................... 17
Tryksøjleindikator ............................................................................ 18
WHO-klassifikationsindikator .......................................................... 18
Om blodtryk .................................................................................... 18
Hvad er blodtryk? ...................................................................... 18
Hvad er hypertension, og hvordan kontrolleres det?................. 19
Hvorfor måle blodtryk derhjemme? ........................................... 19
WHO’s blodtryksklassifikation ................................................... 19
Variationer i blodtryk ................................................................. 19
Fejlfinding ....................................................................................... 20
Vedligeholdelse .............................................................................. 21
Tekniske data ................................................................................. 21
Dansk 1Kære kunde
Tillykke med dit køb af denne A&D blodtryksmåler. Dette apparat er
udviklet med fokus på brugervenlighed og nøjagtighed, og det vil lette
dine daglige aktiviteter omkring blodtryk.
Vi anbefaler, at du gennemlæser denne vejledning omhyggeligt, før
du bruger apparatet for første gang.

Indledende bemærkninger
Denne enhed overholder det europæiske direktiv 93/42 EØF for medicinske
produkter. Dette markeres med-overensstemmelsesmærkatet.
(0123: referencenummeret for det relevante bemyndigede organ)
Apparatet er beregnet til brug på voksne, ikke på nyfødte eller for tidligt
fødte børn.
Anvendelsessteder. Apparatet er beregnet til din egen private brugi hjemmet.
Apparatet er udviklet til at måle blodtryk og pulsfrekvens på mennesker
med henblik på diagnose.

Sikkerhedsforholdsregler
Apparatet er fremstillet med brug af præcisionskomponenter. Store udsving
i temperatur og luftfugtighed samt direkte sollys, stød og støv skal undgås.
Rengør apparatet og manchetten med en tør, blød klud eller en klud, derer fugtet med vand og et neutralt rengøringsmiddel. Brug aldrig sprit,
benzen, fortynder eller andre stærke kemikalier til rengøring af apparatet
eller manchetten.
Undgå at folde manchetten stramt eller opbevare slangen i længere tid,mens den er vredet eller snoet, da det kan forringe levetiden forapparatets dele.
Vær omhyggelig med at forhindre, at spædbørn eller små børn kan blive
kvalt af slangen eller kablet.
Luftslangen må ikke vrides under måling. Det kan medføre personskadepå grund af længerevarende tryk fra manchetten.
Apparatet og manchetten er ikke vandtætte. Undgå, at der kommer regn,
sved eller vand på apparatet og manchetten.
Målingerne kan forstyrres, hvis apparatet anvendes tæt på TV-apparater,
mikrobølgeovne, mobiltelefoner, røntgenapparater eller andre apparater
med stærke elektriske felter.
Trådløse kommunikationsenheder, f.eks. netværksenheder, mobiltelefoner,trådløse telefoner og deres basestationer samt walkie-talkies kan påvirke
Dansk 2

blodtryksmåleren.
Hold derfor en afstand på mindst 30 cm fra sådanne apparater.
Ved genbrug af apparatet skal du kontrollere, at apparatet er rent.
Brugt udstyr, dele og batterier må ikke behandles som almindeligt
husholdningsaffald, men skal bortskaffes i overensstemmelse medgældende lokale regler og bestemmelser.
Når AC-adapteren anvendes, skal du sørge for, at den let kan tages ud af
stikkontakten, når det er nødvendigt.
Der må ikke foretages ændringer af apparatet. Det kan forårsage i ulykker
eller skader på apparatet.
For at måle blodtrykket skal armen klemmes kraftigt nok af manchetten til
midlertidigt at stoppe blodgennemstrømningen gennem arterien. Det kanforårsage smerter, følelsesløshed eller et midlertidig rødt mærke på armen.
Denne tilstand kan især opstå, hvis målingen gentages et antal gange.
Eventuel smerte, følelsesløshed eller røde mærker vil forsvinde med tiden.
Overdrevent hyppig måling af blodtrykket kan forårsage skader, fordi dergribes ind i blodgennemstrømningen. Kontroller, at brugen af apparatet ikke
forringer blodcirkulationen i længere tid, hvis apparatet bruges gentagne
gange.
Hvis du har fået udført en mastektomi (fjernelse af brystet), skal du rådføre dig
med en læge, før du bruger apparatet.
Lad ikke børn bruge apparatet selv, og undgå at bruge apparatet på et sted,hvor små børn har adgang. Det kan forårsage ulykker eller skader.
Apparatet indeholder små dele, som kan udgøre kvælningsfare, hvis etspædbarn sluger en del ved et uheld.
Tag AC-adapteren ud af stikkontakten, når den ikke er i brug undermålingen.
Brug af tilbehør, som ikke er beskrevet i denne manual, kan forringe
sikkerheden.
Hvis batteriet skulle blive kortsluttet, kan det blive varmt og eventueltforårsage forbrændinger.
Lad apparatets temperatur tilpasse sig til omgivelserne før brug (cirka entime).
Der har ikke været udført klinisk test på nyfødte børn og gravide kvinder. Må
ikke bruges på nyfødte spædbørn eller gravide kvinder.
Undgå at røre ved batterierne, DC-stikket og patienten på samme tid. Det
kan medføre elektrisk stød.
Undgå at pumpe op, uden at manchetten er sat omkring overarmen.

Dansk 3

Kontraindikationer
Følgende sikkerhedsforholdsregler skal overholdes for korrekt brug af
apparatet.
Manchetten må ikke sættes på en arm, hvor der er påsat andet medicinsk
elektrisk udstyr. Udstyret vil måske ikke fungere korrekt.
Personer med betydeligt nedsat kredsløb i armen skal rådføre sig med enlæge, før apparatet bruges, for at undgå helbredsproblemer.
Lad være med selv at stille diagnoser ud fra måleresultater og selv starte
en behandling. Kontakt altid din læge med henblik på vurdering af
resultaterne og eventuel behandling.
Manchetten må ikke sættes på en arm, der har et ikke-helet sår.
Manchetten må ikke sættes på en arm, der modtager intravenøst drop
eller blodtransfusion. Det kan medføre personskade eller ulykker.
Apparatet må ikke bruges, hvor der er brændbare gasser, f.eks.
anæstesigasser. Det kan medføre eksplosion.
Apparatet må ikke anvendes i omgivelser med høj iltkoncentration, foreksempel et iltkammer eller et ilttelt. Det kan medføre brand eller
eksplosion.

Dansk 4

Oversigt over deleSkærm

AC-adapterstik

LuftstikDC-stik
Knappen START
Luftfatning

LuftslangeOmråde for korrekt
pasform

Positionsmærke for arterie
Indeksmærke
Arm-manchet
Batterirum
Batteridæksel

1,5 V-batterier(R6P, LR6 eller AA)

SkærmGennemsnit
HUKOMMELSE
Symbol for IHB/AFib%IHB/AFib
Symbol for manchet,der ikke passer

Systolisk tryk

Symbol for bevægelsesfejl

Diastolisk tryk

Hjertemærke
WHO-klassifikationsindikator
og tryksøjleindikator

Pulsfrekvens

Batteriindikator

Dansk 5

Symboler
Symboler, der er trykt på apparatets kabinet
Symboler
SYS
DIA
PUL

IP

Funktion/betydningSæt apparatet på standby og tænd det igen.
Systolisk blodtryk i mmHg
Diastolisk blodtryk i mmHg
Pulsslag pr. minut
Installationsvejledning for batteriJævnstrøm
Type BF: Apparat, manchet og slanger er designet til at givesærlig beskyttelse mod elektrisk stød.
Mærke for EU-direktiv på medicinsk apparat
EU-repræsentant
ProducentFremstillingsdato
Internationalt beskyttelsessymbol
WEEE-mærkat

SN

Serienummer
Se brugermanual/hæfte
Polaritet for DC-stik
Opbevares tørt

Symboler, der vises på skærmen
Symboler

Funktion/betydningAnbefalet handling
Vises, mens måling er i gang. Blinker, Måling er i gang. Undgå
når der registreres puls.
bevægelser.
Symbol for IHB/AFib vises, når derregistreres uregelmæssigt hjerteslag.
Det kan tændes, når der registreresvibrationer i form af skælven ellerrystelse.
Udlæsningen kan haveen forkert værdi. Prøv at
Vises, når der registreres bevægelse
foretage målingen igen.
af kroppen eller armen.
Undgå bevægelser
under målingen.

Dansk 6

Symboler
Symboler

Funktion/betydningAnbefalet handlingUdlæsningen kan have en
forkert værdi. Sæt
Vises under måling, når manchetten
manchetten korrekt på, og
er monteret løst
prøv at foretage målingen
igen.
Registreret IHB/AFib-frekvens i
hukommelsen
Antal registrerede
IHB/AFib’er ihukommelsenx100 [%]%IHB/AFib =Samlet antal
Tidligere målinger lagret i
Hukommelsen.
Gennemsnitsdata
FULDT BATTERI
Indikatoren for batteristrøm undermåling.
LAV BATTERIKAPACITET
Batterikapacitet er lav, når den
blinker.
Ustabilt blodtryk på grund afbevægelse under måling.

Udskift alle batterier mednye, når mærket blinker.
Prøv at foretage målingen
igen.
Man skal undgå atbevæge sig undermålingen.

De systoliske og diastoliske værdier
ligger inden for 10 mmHg fra
hinanden.
Trykværdien steg ikke underSæt manchetten korrekt
oppumpning.
på, og prøv at foretage
målingen igen.
Manchetten er ikke sat korrekt på.
PULSFEJL
Pulsen blev ikke registreret korrekt.
Intern fejl i blodtryksmåleren

Dansk 7

Tag batterierne ud, tryk på
knappen START, og sæt
derefter batterierne i igen.
Kontakt forhandleren, hvis
fejlen stadig vises.

Driftstilstand1. Normal måling
Tryk på knappen START. Blodtrykket måles, og dataene lagres i hukommelsen.
Apparatet kan gemme de sidste 60 målinger i hukommelsen.

2. Hentning af data
Hvis der ikke vises noget, skal du trykke på
knappen START og holde den inde.

I standby:
Tryk på knappen
START og holdden inde

Slip knappen, når gennemsnitsdataene
vises.

Datanummeret og de lagrede data vises
automatisk i rækkefølge fra den senestemåling.

Du kan finde flere oplysninger om at hente
data under “Hentning af hukommelsesdata“.

3. Sletning af alle data, der er lagret i hukommelsen
Tryk på knappen START, mens du slukker apparatet,indtil “” vises.
Vælg “” for at slette alle dataene.
Alle data er slettet når mærket
blinker.
Apparatet slukkes automatisk.

4. Måling med det ønskede systoliske tryk
Se side 15 for måling med det ønskede systoliske tryk.

Dansk 8STARTSådan bruges målerenIsætning/udskiftning af batterier
1. Fjern batteridækslet.

Trin 1
Trin 3

2. Fjern de brugte batterier, og sætnye batterier i batterirummet som
vist. Sørg for, at polerne (+ og -)vender korrekt.
Brug kun batterier af typen R6P,
LR6 eller AA.

Trin 1
Trin 2

3. Sæt batteridækslet på.

FORSIGTIGSæt batterierne i batterirummet som vist. Hvis de sættes forkert i, fungerer
apparatet ikke.
Når(mærke for lavt batteriniveau) blinker på skærmen, skal allebatterierne udskiftes med nye. Brug ikke gamle og nye batterier sammen.
Det kan forkorte batteriets levetid eller få apparatet til at fungere forkert.
Vent mindst to sekunder efter at have slukket apparatet, før du udskifter
batterierne.
Hvis(mærke for lavt batteriniveau) stadig vises, efter at batterierne er
udskiftet, skal du foretage en blodtryksmåling. Derved kan apparatet
registrere de nye batterier.
(mærke for lavt batteriniveau) vises ikke, når batterierne er tomme.
Batteriets levetid afhænger af den omgivende temperatur og kan værekortere ved lave temperaturer. Generelt holder fire nye LR6-batterier cirka
et år ved to daglige målinger.
Brug kun de specificerede batterier. Batterierne, der følger med apparatet,
er til test af målerens funktioner og kan have en begrænset levetid.
Fjern batterierne, hvis apparatet ikke skal bruges i længere tid.
Batterierne kan blive utætte og medføre funktionsfejl.

Dansk 9Sådan bruges måleren
Tilslutning af luftslangen

Luftfatning

Sæt luftstikket godt fast i
luftfatningen.

LuftstikTilslutning af AC-adapterenSæt AC-adapterstikket i
DC-stikket.
Forbind derefter AC-adapteren til
en stikkontakt.

DC-stikAC-adapterstik
Brug den specificerede
AC-adapter. (Se side 22.)Når du tager AC-adapteren ud af stikkontakten, skal du gribe om
AC-adapterens hus og trække det ud af stikkontakten.
Når du tager AC-adapterstikket ud af blodtryksmåleren, skal du gribeom AC-adapterstikket og trække det ud af måleren.

Dansk 10Sådan bruges måleren
Valg af den korrekt manchetstørrelse
En korrekt manchetstørrelse er vigtig for en nøjagtig måling. Hvis
manchetten ikke har den rigtige størrelse, kan målingen vise en forkert
blodtryksværdi.
Armstørrelsen er trykt på hver manchet.
Indeksetog området for korrekt pasform på manchetten fortælle dig, om
du bruger den rigtige manchet. (Se “symboler, der er trykt på manchetten” pånæste side)
Hvis indeksetpeger uden for området, skal du kontakte din lokale
forhandler for at købe en anden manchet.
Arm-manchetten er en forbrugsvare. Køb en ny, hvis det bliver slidt.
Anbefalet
Katalognummer
manchetstørrelse
31 cm til 45 cm
Stor manchet til voksne
CUF-F-LA
22 cm til 42 cm
Universalmanchet
CUF-I
22 cm til 32 cm
Manchet til voksne
CUF-F-A
Armstørrelse: Omkredsen ved biceps.
Bemærk:
UA-767S-W er ikke beregnet til brug af en lille manchet.
ArmstørrelsePåsætning af arm-manchetten
1. Fest manchetten rundt om
overarmen, ca. 1 - 2 cm overindersiden af albuen som vist.
Sæt manchetten direkte mod huden,da tøj kan gøre pulsen svagere, såder opstår fejl i målingen.

Rul ikke skjorteærmet for stramt op.
Manchet1 -2 cm

Luftslange

2. Stramning omkring overarmen på
grund af et stramt oprullet
skjorteærme kan forhindre nøjagtigemålinger.

Positionsmærke for arterie

Klem stoffet fast til.

3. Kontroller, at indeksetpeger
inden for området for korrekt
pasform.
IndeksDansk 11og område for korrekt pasformSådan bruges måleren
Symboler, der er trykt på manchetten
Symboler

Funktion/betydningAnbefalet handlingSæt mærket
på
indersiden af armen ved
arterien i overarmen, ellerså den flugter med
ringfingeren.

Positionsmærke for arterie
▲
REF

Index
KatalognummerOmråde for korrekt pasform formanchet til voksne.
Dette er trykt på manchetten til voksne.
Over området, trykt på manchet til
voksne og universalmanchet.

Brug stor manchet til
voksne i stedet for
manchet til voksne eller
universalmanchet.

Område for korrekt pasform foruniversalmanchet. Dette er trykt påuniversalmanchetten.
Område for korrekt pasform for stor
manchet til voksne. Dette er trykt på denstore manchet til voksne.
Under området, trykt på manchet til
voksne og universalmanchet.
Brug manchet til voksne i
Under området, trykt på stor manchet
stedet for stor manchet til
til voksne.
voksne.
Partinummer
Stor manchet til voksneOmråde for korrekt pasform

A

L

Universalmanchet

S

W

L

A

L

Manchet til voksneS

Dansk 12Sådan bruges målerenSådan foretager du nøjagtige målinger
Gør følgende for at få den mest præcise blodtryksmåling:Sæt dig i en stol, så du sidder behageligt. Hvil armen på bordet. Sid ikke
med krydsede ben. Lad fødderne hvile fladt på gulvet, og ret ryggen.
Slap af i fem til ti minutter før målingen.
Anbring midten af manchetten i samme højde som dit hjerte.
Sid stille, og tal ikke under målingen.
Undlad at måle lige efter fysisk anstrengelse eller badning. Hvil dig i tyve
eller tredive minutter før målingen.
Prøv at måle dit blodtryk på samme tid hver dag.

MålingDet er normalt, at manchetten føles meget stram under målingen (ingen
grund til at være urolig).

Efter måling
Efter målingen skal du trykke på knappen START for at slukke forapparatet.
Fjern manchetten, og skriv dine data ned.
Bemærk: Apparatet har en automatisk strømafbrydelsesfunktion, som afbryder
apparatet cirka et minut efter målingen.
Der skal gå mindst tre minutter mellem målinger på den samme
person.

Dansk 13Målinger
Se “Bemærkninger om nøjagtig måling“ på næste side, før du begynderpå målingen.

Normal måling
1.

Sæt manchetten på armen (helst venstre arm).
Sid stille under målingen.
På højde med hjertet

2.

Tryk på knappen START.
Alle skærmsegmenterne vises.
Tryk på
Nul (0) vises og blinker kortvarigt.
Skærmen skifter som vist i figuren til højre,når målingen starter. Manchetten
Alle skærmsegmenterne
visesbegynder at blive pumpet op. Det er
normalt, at manchetten føles meget stram.
En tryksøjleindikator vises i venstre side afskærmen under oppumpningen.
NulvisningBemærk: Hvis du vil stoppe oppumpningen,Starterkan du til enhver tid trykke på
oppumpningknappen START igen.

3.

Når oppumpningen er færdig, startertømningen af luft automatisk, og(hjertemærke) blinker som angivelse af, atmålingen er i gang. Når pulsen registreres,
blinker mærket med hvert pulsslag.
Bemærk: Hvis der ikke opnås et passende tryk,
starter apparatet automatisk på
oppumpning igen.
Hvis du vil undgå genoppumpning, se“Måling med det ønskede systoliske
tryk“ på næste side.

4.

Når målingen er fuldført, vises de systoliske
og diastoliske trykmålinger samt
pulsfrekvensen.
Manchetten tømmes helt for resterende luft.

5.

Tryk på knappen START for at slukke forapparatet.
Efter et minut uden betjening slukkes apparatet
automatisk.

Bemærk: Der skal gå mindst tre minutter
mellem målinger på den samme
person.
Dansk 14

STARTØger trykket

Måling i gang

Systolisk tryk
Diastolisk tryk
WHO-klassifikation
Pulsfrekvens
Lukker automatisk
resterende luft udMålingerMåling med det ønskede systoliske tryk
UA-767S-W er konstrueret, så den registrerer pulsen og automatisk
pumper manchetten op til systolisk trykniveau.
Brug denne metode, hvis der ofte forekommer genoppumpning, eller hvis
der ikke vises resultater, selvom trykket falder til 20 mmHg eller derunder.
1.

Sæt manchetten på armen i højde med
hjertet (helst venstre arm).

2.

Tryk på knappen START.

3.

Når nullet blinker, skal du trykke på
knappen START og holde den inde,indtil der vises et tal, der er cirka
30 til 40 mmHg højere end det
forventede systoliske tryk.

4.

Når det ønskede tal er nået, skal duslippe knappen START for at startemålingen. Fortsæt for at måle dit
blodtryk som beskrevet på forrige side.

På højde med hjertet
Tryk på START
knappenBemærkninger om nøjagtig måling

Slip knappen ved
det ønskede
systoliske tryk

Tryk på
knappen, og
hold den inde

START

Se den forrige side ommålingSæt dig i en behagelig stilling. Læg armen på et bord med håndfladen
opad og manchetten i samme højde som dit hjerte.
Slap af i fem til ti minutter, før du tager en måling. Hvis du føler diganspændt eller følelsesmæssigt belastet, vil målingen afspejle dettestress som en blodtryksvisning, der er højere (eller lavere) end dennormale blodtryksmåling, og pulsen vil som regel være hurtigere end
normalt.
Dit blodtryk varierer hele tiden, afhængigt af hvad du foretager dig, og hvad
du har spist. Det du drikker, kan have stor og hurtig virkning på dit blodtryk.
Dette apparat baserer sine målinger på pulsen. Hvis du har en meget
svag puls eller uregelmæssig hjerterytme, kan apparatet have svært ved
at finde dit blodtryk.
Hvis apparatet skulle registrere en unormal tilstand, stopper det målingen, og dervises et fejlsymbol. Se side 7 for en beskrivelse af symbolerne.
Dansk 15

Dette apparater er kun beregnet til brug af voksne. Rådfør dig med dinlæge, før du bruger apparatet på et barn. Et barn må ikke bruge apparatet
uden opsyn.
Den automatiske blodtryksmålers ydeevne kan påvirkes af for høj
temperatur, luftfugtighed eller højde over vandets overflade.

Hentning af hukommelsesdataBemærk: Denne enhed lagrer de sidste 60 målinger i hukommelsen.
1.

Hvis der ikke vises noget, skal du
trykke på knappen START og
holde den inde for at hente delagrede data.

2.

Slip knappen, nårgennemsnitsdataene vises.

3.

Datanummeret og de lagrede datavises automatisk i rækkefølge fra
den seneste måling.

Tryk på
og hold

START

knap

Gennemsnit af
systolisk
Gennemsnit af
diastolisk
Gennemsnit af puls
Gennemsnit af
alle dataene

Systolisk

4.

DiastoliskSkærmen slukkes automatisk, efterat alle data er vist.

Puls
Seneste dataBemærk: Hvis du trykker på knappen
START, mens du henter
dataene, slukkes apparatet.

Systolisk
Diastolisk
Puls
Sidste data (ældste)
Dansk 16

Hvad er indikatoren for IHB/AFib?
Hvis måleren registrerer en uregelmæssig rytme under målingerne, vises
indikatoren for IHB/AFib på skærmen med måleværdierne.
Bemærk: Vi anbefaler, at du kontakter din læge, hvis du ofte ser denne
indikator for IHB/AFib.

Hvad er AFib?
Hjertet trækker sig sammen på grund af elektriske signaler i hjertet og
sender blod gennem kroppen. Atrieflimren (AFib) opstår, når det elektriske
signal i hjertets forkammer bliver forstyrret og fører til ændringer
i pulsintervallet. AFib kan få blod til at blive i hjertet, så der let kan dannes
blodpropper, som kan forårsage slagtilfælde og hjerteanfald.

％IHB/AFib%IHB/AFib vises som hyppigheden af den registrerede IHB/AFib.
IHB/AFib kan ikke kun registrere støj som f. eks. fysiske bevægelser, menogså en uregelmæssig hjerterytme. Derfor anbefaler vi, at du kontakter
din læge hvis %IHB/AFib-niveauet er højt.
Antal registrerede
IHB/AFib’er i hukommelsen
x 100 [%]
%IHB/AFib =Samlet antal
Visning af %IHB/AFib: %IHB/AFib vises under visning af gennemsnitsværdier.
(Se ”2. Hentning af data“ i “Driftstilstand“)%IHB/AFib vises ikke, når hukommelsesnummeret er seks
eller derunder.
Visning af gennemsnitsværdi

Niveau 0%IHB/AFib=0

Niveau 1%IHB/AFib=1 - 9

Niveau 2%IHB/AFib=10 - 24

Vises ikke

Dansk 17

Niveau 3%IHB/AFib=25 - 100

Tryksøjleindikator
Indikatoren følger status for trykket under måling.
Øger trykket

Lukker luft ud

Oppumpning i gangMåling i gang
Oppumpning fuldført

WHO-klassifikationsindikator
Hvert segment i søjleindikatoren svarer til
WHO’s blodtryksklassifikation, som er
beskrevet på næste side.
WHO-klassifikationsindikatorSvær hypertension
Moderat hypertension
Mild hypertension
Høj normal
NormalOptimalEksempel:

Moderat hypertension: Indikatoren viser et segment baseret påde aktuelle data, svarende til
WHO-klassifikationen.

Mild hypertension

Høj normal

Om blodtryk
Hvad er blodtryk?
Blodtryk er den kraft, som blodet udøver mod væggene i pulsårerne. Det
systoliske tryk opstår, når hjertet trækker sig sammen. Det diastoliske trykopstår, når hjertet udvider sig. Blodtrykket måles i millimeter kviksølv(mmHg). En persons naturlige blodtryk er det grundlæggende tryk, dermåles som det første om morgenen, mens man stadig er i hvile og ikke
har spist.
Dansk 18

Hvad er hypertension, og hvordan kontrolleres det?
Hypertension, et unormalt højt arterieblodtryk, kan forårsage mange
helbredsproblemer, herunder slagtilfælde og hjerteanfald, hvis det
ignoreres. Hypertension kan kontrolleres ved at ændre livsstil, så manundgår stress, og med medicin under opsyn af en læge.
Sådan kan du forebygge hypertension eller holde den under kontrol:
Ryg ikke
Dyrk regelmæssig motion
Spis mindre salt og fedt
Få dit helbred tjekket regelmæssigt
Hold den rigtige vægtHvorfor måle blodtryk derhjemme?
Blodtryksmåling i en klinik eller hos en læge kan skabe ængstelse og giveen højere måling, 25 til 30 mmHg højere end målt derhjemme. Måling
i hjemmet reducerer udefra kommende påvirkninger af blodtryksmålingen,den supplerer lægens målinger og giver en mere nøjagtig og komplet
historik over blodtrykket.

WHO’s blodtryksklassifikation

Variationer i blodtryk

Referencemateriale: Journal of Hypertension 1999, Vol 17 No.2
Grad 3 hypertension (svær)Diastolisk blodtryk

Verdenssundhedsorganisati
onen (WHO) har udviklet
standarder for vurdering af
forhøjet blodtryk, uafhængigt
af alderen, se diagrammet.

Grad 2 hypertension (moderat)
Grad 1 hypertension (mild)
Høj-normal

Blodtryk

NormalEn persons blodtryk
Optimalvarierer meget i løbet af
dagen og fra årstid til årstid.
Systolisk blodtryk
Det kan variere med
30 til 50 mmHg på grund af forskellige omstændigheder i løbet af dagen.
Hos personer med hypertension er variationerne endnu mere udtalte.
Normalt stiger blodtrykket, når man arbejder eller leger, og falder til det
laveste niveau, når man sover. Bliv derfor ikke unødvendigt bekymret
over resultaterne af en enkelt måling.
Typiske udsving i løbet af dagenTag målingerne på samme tid hver dag,(målt hvert femte minut)og følg proceduren i denne vejledning
Systolisk
for at kende dit normale blodtryk.
Regelmæssige målinger giver en mere
omfattende historik over blodtrykket.
Husk at skrive dato og klokkeslæt, nårdu registrerer dit blodtryk. Rådfør dig
Diastolisk
Søvn
med din læge for at fortolke dineTid
blodtryksdata.

Dansk 19

Fejlfinding
Problem
Der vises intet
på displayet,selvomapparatet ertændt.

Mulig årsag
Batterierne er tømt.

Anbefalet handling
Udskift alle batterierne med nye.

Batteripolerne er ikke
placeret korrekt.

Sæt batterierne i, sådan at denegative og positive poler svarer
til det, der vises i batterirummet.

Batterispændingen er for
lav.
Manchetten(mærke for lavt
pumpes ikke op. batteriniveau) blinker.
Hvis batterierne er tømt
helt, vises mærket ikke.
Manchetten er ikke sat
rigtigt på.
Du har bevæget armeneller kroppen under
Apparatet måler målingen.
ikke.
Målingerne er
Manchetten er ikke
for høje eller for placeret korrekt.
lave.

Udskift alle batterierne med nye.

Sæt manchetten korrekt på.
Sørg for at sidde meget stille, og
tal ikke under målingen.
Sid behageligt og stille.
Læg armen på et bord medhåndfladen opad og manchetten
i samme højde som dit hjerte.
Hvis du har en meget svag puls
eller uregelmæssig hjerterytme,
kan apparatet have svært ved at
finde dit blodtryk.

Værdien er forskellig fra
Se “Hvorfor måle blodtryk
det, der blev målt på enderhjemme?“.
klinik eller hos en læge.
Andet
Tag batterierne ud. Sæt dem
korrekt i, og prøv at foretage
målingen igen.
Bemærk: Hvis de ovenstående handlinger ikke løser problemet, bedes dukontakte forhandleren. Forsøg ikke at åbne eller reparere dette
produkt, da ethvert forsøg på at gøre det vil gøre garantien ugyldig.

Dansk 20

VedligeholdelseUndgå at åbne apparatet. Det bruger følsomme, elektriske komponenterog en kompliceret luftenhed, der kan blive beskadiget. Hvis du ikke kan
løse problemet ved hjælp af vejledning til fejlfinding, bedes du kontakte en
autoriseret forhandler i dit område eller vores kundeserviceafdeling. A&D’s
kundeservice leverer tekniske oplysninger, reservedele og apparater til
autoriserede forhandlere.
Apparatet er designet og fremstillet til at have en lang levetid. Generelt
anbefales det dog at få apparatet efterset hvert 2. år for at sikre korrekt
funktion og nøjagtighed. Kontakt en autoriseret forhandler i dit område
eller A&D for vedligeholdelse.

Tekniske dataTypeMålemetodeMåleområde

UA-767S-W
Oscillometrisk måling
Tryk: 0 - 299 mmHg
Systolisk tryk: 60 - 279 mmHg
Diastolisk tryk: 40 - 200 mmHg
Puls: 40 - 180 slag/minutMålenøjagtighed
Tryk: ±3 mmHg
Puls: ±5 %Strømforsyning
4 x 1,5 V batterier (R6P, LR6 eller AA) eller
AC-adapter (TB-233C) (medfølger ikke)
Antal målinger
Ca. 700 gange LR6 (alkaliske batterier)Ca. 200 gange R6P (manganbatterier)
Ved trykværdi 180 mmHg, stuetemperatur 23 °C.
Klassifikation
Internt strømforsynet udstyr (forsynes fra batterier) /Klasse II (forsynes fra adapter) Kontinuerlig
driftstilstand
Klinisk test
i henhold til ISO81060-2: 2013
I det kliniske valideringsstudie blev K5 anvendt til
85 deltagere for at fastslå det diastoliske blodtryk.
EMD
IEC 60601-1-2: 2014
Hukommelse
Sidste 60 målinger
Driftsforhold+10 til +40 °C / 15 til 85 % RH / 800 til 1060 hPa
Transport / opbevaringsforhold -20 til +60 °C / 10 til 95 % RH / 700 til 1060 hPa
Dimensioner
Ca. 140 [B] x 60 [H] x 105 [D] mm
VægtCa. 245 g uden batterierTæthedsgrad
Apparat: IP20
Dansk 21Påsat delForventet levetid

Manchet Type BF
Apparat: 5 år (ved brug seks gange om dagen)Manchet: 2 år (ved brug seks gange om dagen)AC-adapter: 5 år (ved brug seks gange om dagen)AC adapter (tilbehør)
Adapteren bruges til at forbinde apparatet til en strømkilde i hjemmet.
Kontakt din lokale A&D-forhandler ved køb. AC-adapteren skal efterses eller
udskiftes med jævne mellemrum.
TB-233C
Kontakt din lokale A&D-forhandler vedrørende køb.
AC-adapteren skal efterses eller udskiftes medjævne mellemrum.
Symboler, der er trykt på AC-adapteren
Symboler
Funktion/betydning
Kun til indendørs brug
Klasse II-apparat
Termosikring
SikringMærke for EU-direktiv på apparatet
EAC-certificeringsmærke på apparatet
Polaritet for AC-adapterstik
Tilbehør, som sælges separat
Manchet Katalognummer
Manchetstørrelse
Armstørrelse
CUF-F-LA
Stor manchet til voksne
31 cm til 45 cm
CUF-I
Universalmanchet
22 cm til 42 cm
CUF-F-A
Manchet til voksne
22 cm til 32 cm
Armstørrelse: omkredsen ved biceps.
AC-adapter

Katalognummer
TB-233C

Stik (stikkontakttype)Type CBemærk: Specifikationerne kan ændres uden forudgående varsel.
IP-klassificering er beskyttelsesgraden, som ydes af kabinetter ioverensstemmelse med IEC 60529. Dette apparat er beskyttet mod
faste fremmedlegemer på 12 mm i diameter og derover, f.eks. fingre.
Denne enhed er ikke beskyttet mod vand.

Dansk 22

Innehåll
Bästa kund ........................................................................................ 2
Inledande anmärkningar ................................................................... 2
Försiktighetsåtgärder ........................................................................ 2Ingående delar .................................................................................. 5
Symboler........................................................................................... 6
Driftsläge........................................................................................... 8
1. Normal mätning ....................................................................... 82. Hämta värden .......................................................................... 8
3. Radera alla värden som sparats i minnet ................................ 8
4. Mätning med önskat systoliskt tryck ........................................ 8Använda mätaren ............................................................................. 9
Installera/byta batterier ................................................................ 9
Ansluta luftslangen .................................................................... 10
Ansluta nätadaptern .................................................................. 10
Välja rätt manschettstorlek ........................................................ 11
Sätta på armmanschetten ......................................................... 11Så här görs korrekta mätningar ................................................. 13
Mätning ..................................................................................... 13
Efter mätningen ......................................................................... 13
Mätningar ........................................................................................ 14
Normal mätning ......................................................................... 14
Mätning med önskat systoliskt tryck .......................................... 15
Anmärkningar för korrekta mätningar ........................................ 15
Hämta värden ur minnet ................................................................. 16
Vad är IHB/AFib-indikatorn? ........................................................... 17
Vad är AFib (förmaksflimmer)? ....................................................... 17%IHB/AFib (arytmi/förmaksflimmer) ............................................... 17
Tryckindikator ................................................................................. 18
Indikator för WHO-klassificering ..................................................... 18
Om blodtryck................................................................................... 18
Vad är blodtryck? ...................................................................... 18
Var är hypertoni och hur kontrolleras det? ................................ 19
Varför mäta blodtrycket i hemmet? ........................................... 19
WHO:s blodtrycksklassificering ................................................. 19
Variationer i blodtrycket ............................................................. 19
Felsökning ...................................................................................... 20
Underhåll ........................................................................................ 21
Tekniska data ................................................................................. 21
Svenska 1

Bästa kund
Gratulerar till ditt inköp av en toppmodern blodtrycksmätare från A&D.
Den här enheten är utformad för att vara enkel att använda och samtidigt
vara noggrann och underlättar din dagliga blodtrycksmätningsrutin.
Vi rekommenderar att du läser igenom den här bruksanvisningen
noggrant innan du använder produkten för första gången.

Inledande anmärkningar
Den här enheten överensstämmer med Rådets direktiv 93/42/EEG om
medicintekniska produkter. Det synliggörs genomöverensstämmelsemärkningen. (0123: Referensnumret avser det
anmälda organ som är involverat)
Enheten är utformad för användning på vuxna, inte på nyfödda ellersmåbarn.
Användningsmiljö. Enheten är avsedd för personligt bruk för vård i hemmet.
Den här enheten är avsedd för att mäta blodtrycket och pulsfrekvensen
hos människor i diagnostiskt syfte.

Försiktighetsåtgärder
Precisionskomponenter har använts vid tillverkningen av den här enheten.
Undvik extrema temperaturer och luftfuktighet, direkt solljus, stötar och
damm.
Rengör enheten och manschetten med en torr, mjuk trasa eller en trasa
fuktad med vatten och ett neutralt rengöringsmedel. Använd aldrig
alkohol, bensen, thinner eller andra kraftiga kemikalier för att rengöra
enheten eller manschetten.
Undvik att vika ihop manschetten för hårt och att förvara slangen hårtlindad under längre tidsperioder eftersom det kan förkorta
komponenternas livslängd.
Vidta åtgärder så att det inte finns risk för att spädbarn och små barn
stryps av slangen och kabeln.
Vrid inte luftslangen under mätningen. Det kan orsaka personskada på
grund av det kontinuerliga manschettrycket.
Enheten och manschetten är inte vattentäta. Undvik regn, svett ochvatten på enheten och manschetten.
Mätningarna kan bli felaktiga om enheten används i närheten av
TV-apparater, mikrovågsugnar, mobiltelefoner eller röntgenutrustningeller andra enheter med kraftiga magnetfält.
Trådlösa kommunikationsenheter, t.ex. hemnätverksenheter, mobiltelefoner,trådlösa telefoner och deras basstationer samt walkie-talkies kan påverka
den här blodtrycksmätaren.
Därför ska sådana enheter hållas på ett avstånd om 30 cm.
Kontrollera att enheten är ren när den återanvänds.
Svenska 2

Använd utrustning, delar och batterier betraktas inte som vanligthushållsavfall och måste kasseras i enlighet med lokala föreskrifter.
När nätadaptern används, kontrollera att nätadaptern enkelt kan dras ur
eluttaget vid behov.
Gör inga förändringar på enheten. Det kan orsaka olyckor eller skada
enheten.
För att mäta blodtrycket måste armen tryckas ihop av manschetten
tillräckligt hårt för att blodflödet genom artären ska stoppas tillfälligt. Det
kan orsaka smärta, domningar eller ett tillfälligt rött märke på armen.
Detta tillstånd uppträder särskilt vid upprepade på varandra följande
mätningar. All smärta, domningar eller röda märken försvinner med tiden.
Om blodtrycket mäts för ofta kan det leda till skada på grund av störningar
i blodflödet. Kontrollera att användningen av enheten inte leder tilllångvarig försämring av blodcirkulationen när enheten används
upprepade gånger.
Om du genomgått mastektomi ska du kontakta läkare innan du använder
den här enheten.
Låt inte barn använda den här enheten själva och använd inte den här
enheten inom räckhåll för småbarn. Det kan orsaka olyckor eller skador.
Det finns små delar som kan leda till risk för kvävning om små barnsväljer dem av misstag.
Koppla bort nätadaptern när den inte används under mätningen.
Användning av tillbehör som inte beskrivs i den här bruksanvisningen kanäventyra säkerheten.
Om batteriet kortsluts kan det bli varmt och möjligen orsaka brännskador.
Låt enheten anpassa sig till omgivningen innan den används (ungefär entimme).
Kliniska prövningar har inte genomförts på nyfödda barn och gravida
kvinnor. Får inte användas på nyfödda barn eller gravida kvinnor.
Vidrör inte batterierna, likströmsuttaget och patienten samtidigt. Det kan
leda till elektriska stötar.
Blås inte upp manschetten om den inte sitter runt överarmen.

Kontraindikationer
Följande är försiktighetsåtgärder för korrekt användning av enheten.
Sätt inte manschetten på en arm som är ansluten till annan medicinsk
elektrisk utrustning. Utrustningen kanske inte fungerar korrekt.
Personer med allvarlig cirkulationsrubbning i armen måste kontakta en
läkare före användning av enheten för att undvika medicinska problem.
Ställ ingen diagnos själv baserat på mätresultaten och starta ingen
behandling själv. Kontakta alltid läkare för en bedömning av resultaten
och behandlingen.
Sätt inte manschetten på en arm med ett sår som inte läkt.
Sätt inte manschetten på en arm som används för intravenöst dropp eller
Svenska 3

blodtransfusion. Det kan orsaka skador eller olyckor.
Använd inte enheten i närheten av lättantändliga gaser som narkosgaser.
Det kan orsaka explosion.
Använd inte enheten i omgivningar med hög koncentration av syrgas, t.ex.
syrgaskammare med högtryck eller syrgastält. Det kan orsaka brand eller
explosion.

Svenska 4Ingående delar
Skärm

Luftkoppling

Kontakt för nätadapter

Likströmsuttag

START -knapp
LuftanslutningOmfång för
korrekt passform

Luftslang
Positioneringsmärke för artär

Indexmärke
ArmmanschettBatterifack
Batterilucka

1,5 V batterier(R6P, LR6 eller AA)

Skärm

Genomsnitt

MINNE
Symbol för IHB och AFib
Systoliskt
tryck%IHB/AFib
Symbol för fel manschettstorlek

Diastoliskt
tryck

Symbol för rörelsefel
Hjärtsymbol
Indikator för WHO-klassificeringoch tryckindikator

Pulsfrekvens

Batteriindikator
Svenska 5

Symboler
Symboler tryckta på enhetens hölje
Symboler
SYS
DIA
PUL

IP

Funktion/innebörd
Standby och slå på enheten.
Systoliskt blodtryck i mmHg
Diastoliskt blodtryck i mmHg
Puls per minut
Guide för installation av batteriet
Likström
Typ BF: Enheten, manschetten och slangen är utformade så att
de ger särskilt skydd mot elektriska stötar.
Märkning för medicinteknisk produkt enligt EU-direktivet
EU-representant
Tillverkare
Tillverkningsdatum
Symbol för internationell skyddsklass
WEEE-märkning

SN

Serienummer
Se bruksanvisningen
Likströmsuttagets polaritet
Förvaras torrt

Symboler som visas på skärmen
Symboler

Funktion/innebörd
Rekommenderad åtgärd
Visas när mätning pågår. Blinkar när Mätning pågår. Var så
pulsen identifieras.
stilla som möjligt.
Symbol för arytmi (IHB) och
förmaksflimmer (AFib)
Visas när en oregelbunden hjärtrytm
identifieras.
Kan tändas när en mycket liten
vibration som en rysning eller skakning
identifieras.
Mätningen kan ge ett
Visas när arm- eller kroppsrörelse
felaktigt värde. Mät igen.
identifieras.
Var stilla undermätningen.

Svenska 6

Symboler
Symboler

Funktion/innebörd

Rekommenderad åtgärd
Mätningen kan ge ett
felaktigt värde. Sätt på
manschetten korrekt och
mät igen.

Visas under mätningen när
manschetten sitter löst
Identifierad frekvens av IHB/AFib iminnet%IHB/AFib =Antal identifierade
IHB/AFib i minnet

x100 [%]

Totalt antal

Tidigare mätningar sparade i
MINNET.
Genomsnittsvärde
FULLT BATTERIBatteriströmindikatorn undermätning.
SVAGT BATTERIBatteriet är svagt när symbolen
blinkar.
Ojämnt blodtryck på grund av
rörelser under mätningen.
De systoliska och diastoliska
värdena ligger inom 10 mmHg frånvarandra.
Tryckvärdet ökade inte under
uppblåsningen.

Byt ut alla batterier mot nya
när symbolen blinkar.
Mät igen.
Var mycket stilla undermätningen.

Sätt på manschetten
korrekt och mät igen.

Manschetten sitter inte korrekt.
PUL-FEL
Pulsen identifierades inte korrekt.
Internt fel i blodtrycksmätaren

Svenska 7

Ta ut batterierna, tryck på
knappen START o ch sättin batterierna igen. Om
felet kvarstår, kontaktaåterförsäljaren.

Driftsläge
1. Normal mätning
Tryck på knappen START . Blodtrycket mäts och data sparas i minnet. Den
här enheten kan spara de senaste 60 mätningarna i minnet.

2. Hämta värden
När ingenting visas, håll in knappen START .

I standby-läget,håll inSTART

knappenSläpp knappen när genomsnittsvärdet visas.

Värdets nummer och sparade värden visas
automatiskt sorterade från den senaste
mätningen.

Information om hur man hämtar värden finns
i “Hämta värden ur minnet“.

3. Radera alla värden som sparats i minnet
När enheten stängs av, håll in knappen
START tills “” visas.
Välj “” för att radera värdena.
Värdena rensas när
blinkar.
Enheten stängs av automatiskt.

4. Mätning med önskat systoliskt tryck
På sida 15 visas hur man mäter med ett önskat systoliskt tryck.

Svenska 8

START

Använda mätaren
Installera/byta batterier

Steg 1

1. Ta bort batteriluckan.
2. Ta bort de förbrukade batterierna
och sätt in nya batterier ibatterifacket som på bilden. Varnoga med att polerna (+ och -)sitter korrekt.
Använd endast batterier av typen
R6P, LR6 eller AA.

Steg 3

Steg 1
Steg 2

3. Sätt på batteriluckan.

VAR FÖRSIKTIG
Sätt in batterierna i batterifacket som på bilden. Om de sätts in på fel sättfungerar inte enheten.
När(SVAGT BATTERI) blinkar på skärmen ska alla batterier bytas ut
mot nya. Blanda inte gamla och nya batterier. Det kan förkorta batteriernas
livslängd eller göra så att enheten fungerar på fel sätt.
Vänta i två sekunder eller mer efter att enheten stängts av innan du byter
batterierna.
Om(SVAGT BATTERI) visas även efter att batterierna bytts ut ska
du göra en blodtrycksmätning. Enheten kan då känna igen de nya
batterierna.
(SVAGT BATTERI) visas inte när batterierna är slut.
Batteriernas livslängd varierar med omgivningstemperaturen och kan vara
kortare vid låga temperaturer. I allmänhet räcker fyra nya LR6-batterier
ungefär ett år vid mätning två gånger per dag.
Använd endast angivna batterityper. Batterierna som medföljer enheten ärtill för att testa mätarens funktion och kan ha en begränsad livstid.
Ta ut batterierna om enheten inte ska användas under en längre tid.
Batterierna kan läcka och orsaka felfunktion.

Svenska 9

Använda mätaren
Ansluta luftslangen

Luftanslutning

Sätt in luftkopplingen korrekt iluftanslutningen.

Luftkoppling

Ansluta nätadapternSätt i kontakten för nätadaptern ilikströmsuttaget.
Anslut sedan nätadaptern till ett
eluttag.

Likströmsuttag

Kontakt för nätadapter
Använd den angivna
nätadaptern. (Se sida 22.)
När nätadaptern kopplas bort från eluttaget, fatta tag i nätadaptern och
dra ut den ur eluttaget.
När kontakten för nätadaptern kopplas bort från blodtrycksmätaren,
fatta tag i kontakten för nätadaptern och dra ut den ur mätaren.

Svenska 10

Använda mätaren
Välja rätt manschettstorlek
Det är viktigt att använda rätt manschettstorlek för en korrekt mätning. Om
manschetten inte har rätt storlek kan mätningen ge ett felaktigt
blodtrycksvärde.
Armstorleken är tryckt på varje manschett.
Indexetoch omfång för korrekt passform på manschetten, visar om duanvänder korrekt manschett. (Se “Symboler tryckta på manschetten” pånästa sida)
Om indexet
pekar utanför omfånget, kontakta den lokalaåterförsäljaren för att köpa en utbytesmanschett.
Armmanschetten är en förbrukningsvara. Köp en ny om den blir sliten.
Rekommenderad
Katalognummer
manschettstorlek
31 cm till 45 cm
Stor vuxenmanschett
CUF-F-LA
Manschett med stort
22 cm till 42 cm
CUF-Iomfång22 cm till 32 cm
Vuxenmanschett
CUF-F-A
Armstorlek: Omkretsen vid biceps.
Obs!
UA-767S-W är inte avsedd att användas med en liten
manschett.
Armstorlek

Sätta på armmanschetten
1. Lägg manschetten runt överarmen,
ungefär 1–2 cm över armbågens
insida som på bilden.
Placera manschetten direkt mot
huden eftersom klädesplagg kan
göra att pulsen blir svag och leda till
felaktig mätning.

Rulla inte upp ärmen hårt.
Manschett
1 -2 cm

Luftslang

Positioneringsmärke för artär

2. Åtdragning runt överarmen på grund
av att en skjortärm rullats upp kan
förhindra korrekta värden.

Fäst tygfästet
ordentligt.

3. Bekräfta att indexet
pekar inomomfånget för korrekt passform.
Index

Svenska 11

och omfång för korrekt passformAnvända mätaren
Symboler tryckta på manschetten
Symboler

Funktion/innebörd

Rekommenderad åtgärd
Placera märket
påöverarmens artär eller ilinje med ringfingret påarmens insida.

Positioneringsmärke för artär
▲
REF

Index
KatalognummerOmfång för korrekt passform för
vuxenmanschetten.
Står tryckt på vuxenmanschetten.
Använd den storaÖver omfånget som är tryckt påvuxenmanschetten istället
vuxenmanschetten och manschetten för vuxenmanschetten
med stort omfång.
eller manschetten med
stort omfång.
Omfång för korrekt passform för
manschetten med stort omfång. Stårtryckt på manschetten med stortomfång.
Omfång för korrekt passform för den
stora vuxenmanschetten. Står trycktpå den stora vuxenmanschetten.
Under omfånget som är tryckt påvuxenmanschetten och manschetten
med stort omfång.
Använd vuxenmanschetten
Under omfånget som är tryckt på den
istället för den stora
stora vuxenmanschetten.
vuxenmanschetten.
Lotnummer

Stor vuxenmanschettAOmfång för korrekt passform

L

Manschett med stort omfång

S

W

L

A

L

VuxenmanschettS

Svenska 12

Använda mätarenSå här görs korrekta mätningar
För noggranna blodtrycksmätningar:
Sitt bekvämt på en stol. Vila armen på bordet. Korsa inte benen. Hållfötterna platt mot golvet och räta på ryggen.
Koppla av i ungefär fem till tio minuter före mätningen.
Placera manschettens mitt i jämnhöjd med hjärtat.
Var stilla och tyst under mätningen.
Mät inte direkt efter fysisk aktivitet eller bad. Vila i tjugo eller trettio minuterföre mätningen.
Försök att mäta blodtrycket vid samma tidpunkt varje dag.

Mätning
Under mätningen är det normalt att manschetten känns väldigt trång.
(Bli inte orolig).

Efter mätningen
Tryck på knappen START för att stänga av enheten efter mätningen.
Ta bort manschetten och anteckna dina värden.
Obs! Enheten har en automatisk avstängningsfunktion som stänger av
enheten ungefär en minut efter mätningen.
Låt minst tre minuter gå mellan mätningar på samma person.

Svenska 13

Mätningar
Före mätning, se “Anmärkningar för korrekta mätningar“ på nästa sida.

Normal mätning
1.

2.

Placera manschetten på armen(helst vänster arm).
Var tyst under mätningen.

I höjd med
hjärtatTryck på knappen START .
Alla skärmens segment visas.
Noll (0) blinkar under en kort tid.
Skärmen ändras, som visas i figuren till
höger, när mätningen börjar. Manschettenbörjar blåsas upp. Det är normalt att
manschetten känns väldigt trång. En
tryckindikator visas till vänster på skärmen
under uppblåsningen.
Obs! Om du vill avbryta uppblåsningen när
som helst, tryck på knappen START
igen.

Tryck
Alla skärmens
segment visas

Visar nollStartaruppblåsning3.

När uppblåsningen är klar startar
tömningen automatiskt och(hjärtsymbol) blinkar, vilket indikerar att
mätningen pågår. När pulsen identifieras
blinkar symbolen vid varje pulsslag.
Obs! Om ett korrekt tryck inte erhålls startar
enheten uppblåsningen igen automatiskt.
För att undvika upprepad uppblåsning, se“Mätning med önskat systoliskt tryck“ pånästa sida.

4.

När mätningen är klar visas det systoliska
och diastoliska trycket samt
pulsfrekvensen.
Manschetten släpper ut kvarvarande luft
och töms helt.

5.

Tryck på knappen START för att stänga
av enheten.
När enheten inte använts på en minut
stängs den av automatiskt.

Obs! Låt minst tre minuter gå mellan
mätningar på samma person.
Svenska 14

START

Trycksätter

Mätning pågår

Systoliskt tryck
Diastoliskt tryck
WHO-klassificering
Pulsfrekvens

Driver ut kvarvarande luft
automatisktMätningar
Mätning med önskat systoliskt tryck
UA-767S-W är utformad så att den identifierar pulsen och blåser upp
manschetten till ett systoliskt tryck automatiskt.
Använd den här metoden när manschetten blåser upp sig upprepadegånger eller när resultaten inte visas även om trycket minskar till
20 mmHg eller lägre.
1.

Placera manschetten på armen i höjd med I höjd med hjärtat
hjärtat (helst vänster arm).

2.

Tryck på knappen START .

3.

När nollan blinkar, håll in knappen
START tills ett värde som är cirka 30
till 40 mmHg högre än ditt förväntade
systoliska tryck visas.

Släpp knappen vid
det önskade
systoliska trycket4.

När det önskade värdet nåtts, släpp
knappen START för att starta
mätningen. Fortsätt att mäta dittblodtryck enligt beskrivningen på
föregående sida.

Håll in knappen

Tryck på START
knappenAnmärkningar för korrekta mätningar

START

Se föregående sida för
mätning

Sitt i en bekväm position. Lägg armen på ett bord med handflatan uppåt
och manschetten i höjd med hjärtat.
Koppla av i ungefär fem till tio minuter innan du mäter. Om du är
upphetsad eller nedstämd av emotionell stress visar mätningen den
stressen som ett högre (eller lägre) blodtrycksvärde än normalt och
pulsvärdet kommer vanligtvis att vara högre än normalt.
En persons blodtrycksvärde varierar kontinuerligt beroende på vad du gör
och vad du har ätit. Vad du dricker kan ha en mycket stor och snabb
effekt på blodtrycket.
Den här enheten baserar mätvärdena på hjärtslagen. Om du har mycket
svaga eller oregelbundna hjärtslag kan enheten att ha svårt att bestämma
ditt blodtryck.
Om enheten identifierar ett onormalt tillstånd avbryts mätningen och en
felsymbol visas. På sida 7 finns en förklaring till symbolerna.
Svenska 15

Den här enheten är avsedd att användas av vuxna. Kontakta läkare innanden här enheten används på barn. Barn ska inte använda den här
enheten utan övervakning.
Den automatiska blodtrycksmätarens funktion kan påverkas av för hög
temperatur, luftfuktighet eller altitud.

Hämta värden ur minnet
Obs! Den här enheten kan spara de senaste 60 mätningarna i minnet.
1.

När ingenting visas, håll in
knappen _START_ för att hämta
de sparade värdena.

2.

Släpp knappen när
genomsnittsvärdet visas.

3.

Värdets nummer och sparade
värden visas automatiskt sorteradefrån den senaste mätningen.

4.

Tryck och START knappenhåll in
Genomsnittligt
systoliskt
Genomsnittligt
diastoliskt
Genomsnittlig puls

Genomsnitt för
alla värden

Skärmen stängs av automatiskt efter
att alla värden visats.

Systoliskt
Diastoliskt
Puls
Senaste värde

Obs! Om du trycker på knappen_START_ medan värden hämtas
stängs enheten av.

Systoliskt
Diastoliskt
Puls
Sista värdet (äldst)
Svenska 16

Vad är IHB/AFib-indikatorn?
När mätaren identifierar en oregelbunden frekvens under mätningenvisas IHB/AFib-indikatorn på skärmen med mätvärdena.
Obs! Vi rekommenderar att du kontaktar läkare om du ser den här
IHB/AFib-indikatorn ofta.

Vad är AFib (förmaksflimmer)?
Hjärtat drar ihop sig på grund av elektriska signaler i hjärtat och skickar ut
blod genom kroppen. Förmaksflimmer (AFib) inträffar när den elektriska
signalen i förmaket blir störd och leder till störningar i pulsintervallet.
Förmaksflimmer (AFib) kan leda till att blod stannar i hjärtat och kan lättbilda koagel, vilket är en orsak till stroke och hjärtattack.

%IHB/AFib (arytmi/förmaksflimmer)%IHB/AFib visas som frekvensen för att identifiera IHB/AFib(arytmier/förmaksflimmer).
IHB/AFib kan inte bara identifiera brus, som fysisk rörelse, utan även ett
oregelbundet hjärtslag. Därför rekommenderar vi att du kontaktar läkare
om %IHB/AFib är högt.
Antal identifierade
IHB/AFib i minnet
x 100 [%]
%IHB/AFib =Totalt antal
Visning av %IHB/AFib: %IHB/AFib visas när genomsnittsvärden visas.
(Se “2. Hämta värden“ i “Driftsläge“)%IHB/AFib visas inte när antalet i minnet är sex
eller lägre.
Visning av genomsnittliga värden

Nivå 0%IHB/AFib=0Nivå 1Nivå 2Nivå 3%IHB/AFib=1–9 %IHB/AFib=10–24 %IHB/AFib=25–100

Visas inte

Svenska 17

Tryckindikator
Indikatorn visar tryckets förlopp under mätningen.
Trycksätter

Släpper ut luft

Uppblåsning pågår

Mätning pågår
Uppblåsning klar

Indikator för WHO-klassificering
Varje segment i indikatorn motsvarar
WHO:s blodtrycksklassificering enligt
beskrivningen på nästa sida.

Exempel:Indikator för WHO-klassificering
Allvarlig hypertoniMåttlig hypertoni
Lindrig hypertoni
Hög normal
NormalOptimal: Indikatorn visar ett segment, baserat på
aktuella värden, som motsvarar WHO:sklassificering.

Måttlig hypertoni

Lindrig hypertoni

Hög normal

Om blodtryck
Vad är blodtryck?
Blodtryck är den kraft som hjärtat utövar mot artärväggarna. Systoliskt
tryck inträffar när hjärtat drar ihop sig. Diastoliskt tryck inträffar när hjärtat
utvidgar sig. Blodtrycket mäts i millimeter kvicksilver (mmHg). En persons
naturliga blodtryck representeras av det grundläggande trycket som mäts
direkt på morgonen när man är utvilad och innan måltid.

Svenska 18

Var är hypertoni och hur kontrolleras det?
Hypertoni – ett onormalt högt artärblodtryck – kan, om det inte behandlas,orsaka flera hälsoproblem inklusive stroke och hjärtattack. Hypertoni kan
kontrolleras genom att ändra livsstil, undvika stress och med läkemedel
under läkares övervakning.
För att förhindra hypertoni eller hålla det under kontroll:
Rök inte
Motionera regelbundet
Minska intaget av salt och fett
Gör regelbundna hälsokontrollerHåll en lämplig vikt

Varför mäta blodtrycket i hemmet?
När blodtrycket mäts på klinik eller läkarmottagning kan det skapaängslan som kan föranleda ett förhöjt värde, 25 till 30 mmHg högre än det
som mäts i hemmet. Mätning i hemmet minskar effekterna av yttrepåverkan på blodtrycksvärdena, utgör ett komplement till läkarens
mätningar och ger en mer exakt och fullständig blodtryckshistorik.

WHOs blodtrycksklassificering

Variationer i blodtrycket

Referensmaterial: Journal of Hypertension 1999, Vol 17 No.2
Hypertoni grad 3 (allvarlig)
Diastoliskt blodtryck

Standarder för bedömningav högt blodtryck, oavsettålder, har upprättats av
Världshälsoorganisationen(WHO), som visas i
diagrammet.

Hypertoni grad 2 (måttlig)
Hypertoni grad 1 (lindrig)
Hög-normal
NormalOptimalBlodtryck

En persons blodtryck
varierar mycket dag till dag
Systoliskt blodtryck
och beroende på årstid. Det
kan variera med 30 till 50 mmHg på grund av olika förutsättningar underdagen. Hos personer med hypertoni är variationerna ännu större.
Normalt stiger blodtrycket vid arbete eller lek och sjunker till sin lägsta
nivå när man sover. Bli därför inte alltför oroad över resultaten från en
mätning.
Typiska fluktuationer under en
Mät vid samma tid varje dag meddag (mätt var femte minut)
metoden som beskrivs i den här
Systoliskt
bruksanvisningen så att du lär känna dittnormalablodtryck.
Regelbundna
mätningar ger en mer fullständig
blodtryckshistorik. Var noga med att
notera datum och tid när du mäter dittDiastoliskt
Sova
blodtryck. Kontakta läkare för att tolka dinaTid
blodtrycksvärden.

Svenska 19

Felsökning
Problem
Ingenting visas
på skärmen,även när
enheten ärpåslagen.

Möjlig orsak
Batterierna är slut.

Rekommenderad åtgärd
Byt ut alla batterierna mot nya.
Sätt in batterierna så att den
Batteriernas poler är inte negativa och den positiva polen
i rätt position.
matchar de som visas ibatterifacket.
Batterispänningen är för
låg.
Manschetten(SVAGT BATTERI)
Byt ut alla batterierna mot nya.
blåses inte upp. blinkar. Om batterierna
är helt slut visas inte
symbolen.
Manschetten sitter inte
Sätt på manschetten korrekt.
korrekt.
Du flyttade armen eller
Var helt stilla och tyst underkroppen undermätningen.
Enheten mäter mätningen.
inte.
Sitt bekvämt och stilla.
Värdena är för Manschettens position
Lägg armen på ett bord med
höga eller för
är inte korrekt.
handflatan uppåt ochlåga.
manschetten i höjd med hjärtat.
Om du har mycket svaga eller
oregelbundna hjärtslag kan
enheten att ha svårt att
bestämma ditt blodtryck.
Värdet skiljer sig fråndet som uppmätts på en Se “Varför mäta blodtrycket i
klinik eller
hemmet?“.
läkarmottagning.
ÖvrigtTa ut batterierna. Sätt tillbaka
dem korrekt och försök att mäta
igen.
Obs! Om åtgärderna som beskrivs ovan inte löser problemet, kontaktaåterförsäljaren. Försök inte att öppna eller reparera den här produkten– garantin gäller då inte längre.

Svenska 20

Underhåll
Öppna inte enheten. Den har ömtåliga elektriska komponenter och en
invecklad luftenhet som kan skadas. Om problemet inte kan lösas med
felsökningsanvisningarna, kontakta den auktoriserade återförsäljaren i dittområde eller vår kundtjänst. Kundtjänsten hos A&D ger teknisk information,
tillhandahåller reservdelar och enheter till auktoriserade återförsäljare.
Enheten har utformats och tillverkats för att få en lång livslängd. Vi
rekommenderar emellertid normalt att låta inspektera enheten vartannat
år för att säkerställa korrekt funktion och noggrannhet. Kontakta den
auktoriserade återförsäljaren i ditt område eller A&D för underhåll.

Tekniska data
Typ
Mätmetod
Mätintervall

Mätnoggrannhet
Strömförsörjning
Antal mätningar

Klassificering

Kliniskt test

EMD
Minne
Driftförhållanden
Transport-/förvaringsförhållanden
Mått
Vikt
Kapslingsklass

UA-767S-W
Oscillometrisk mätning
Tryck:
0–299 mmHg
Systoliskt tryck: 60–279 mmHg
Diastoliskt tryck: 40–200 mmHg
Puls:
40–180 slag/minut
Tryck:±3 mmHgPuls:±5 %4 x 1,5 V batterier (R6P, LR6 eller AA) eller
nätadapter (TB-233C) (medföljer inte)
Cirka 700 gånger
LR6 (alkaliska batterier)Cirka 200 gånger
R6P (manganbatterier)
Med tryckvärde 180 mmHg, rumstemperatur 23 °C.
Internt strömförsörjd medicinsk elektrisk utrustning(drivs med batterier)/klass II(drivs med nätadapter)
Kontinuerligt driftläge
I enlighet med ISO 81060-2:2013
I den kliniska valideringsstudien användes K5 på
85 subjekt för bestämning av diastoliskt blodtryck.
IEC 60601-1-2: 2014
Senaste 60 mätningarna+10 till +40 °C/15 till 85 %RH/800 till 1060 hPa-20 till +60 °C/10 till 95 %RH /700 till 1060 hPa
Cirka 140 [B] x 60 [H] x 105 [D] mm
Cirka 245 g, exklusive batterierna
Enhet:
IP20
Svenska 21

Applicerad delAnvändbar livstid

Manschett
Typ BF
Enhet: 5 år (vid användning sex gånger per dag)
Manschett: 2 år (vid användning sex gånger per
dag)
Nätadapter: 5 år (vid användning sex gånger per
dag)

Nätadapter som tillbehör
Adaptern är avsedd för att ansluta enheten till en strömkälla i hemmet.
Kontakta den lokala återförsäljaren för A&D för inköp. Nätadaptern måste
inspekteras eller bytas ut regelbundet.
TB-233C
Kontakta den lokala återförsäljaren för A&D för
inköp.
Nätadaptern måste inspekteras eller bytas ut
regelbundet.
Symboler tryckta på nätadaptern
Symboler
Funktion/innebörd
Endast för inomhusbruk
Enhet av klass II
Termisk säkring
Säkring
Märkning för medicinteknisk produkt
enligt EU-direktivet
Märkning för certifiering enligt EAC
Polaritet för kontakten för nätadaptern
Tillbehör säljs separat
Manschett
Katalognummer
CUF-F-LA

Manschettstorlek
Stor vuxenmanschett
Manschett med stort
CUF-Iomfång
CUF-F-A
VuxenmanschettArmstorlek: Omkretsen vid biceps.

Nätadapter
Obs!

Katalognummer
TB-233C

Armstorlek
31 cm till 45 cm
22 cm till 42 cm
22 cm till 32 cm

Kontakt (för vägguttag)
Typ CSpecifikationer kan ändras utan föregående meddelande.
IP-klassningen är den nivå av skydd som inkapslingarna ger enligt
IEC 60529. Den här enheten är skyddad mot fasta främmande
föremål med 12 mm diameter och större, t.ex. ett finger. Denna enhet
är inte skyddad mot vatten.

Svenska 22

InnholdKjære kunder .................................................................................... 2
Innledende merknader ...................................................................... 2
Forholdsregler................................................................................... 2
Identifisering av deler........................................................................ 5
Symboler........................................................................................... 6
Driftsmodus....................................................................................... 8
1. Normal måling ......................................................................... 8
2. Hente frem data ...................................................................... 8
3. Slette alle lagrede verdier i minnet .......................................... 8
4. Målinger med ønsket systolisk trykk........................................ 8
Bruke måleren .................................................................................. 9Sette inn/skifte batterier .............................................................. 9
Koble til luftslangen ................................................................... 10
Koble til strømadapteren ........................................................... 10
Velge riktig mansjettstørrelse .................................................... 11
Sette på mansjetten .................................................................. 11
Foreta nøyaktige målinger ........................................................ 13Måling........................................................................................ 13
Etter målingen ........................................................................... 13Målinger .......................................................................................... 14
Normal måling ........................................................................... 14Målinger med ønsket systolisk trykk.......................................... 15
Merknader for nøyaktig måling .................................................. 15
Hente frem data fra minnet ............................................................. 16
Hva er IHB/AFib-indikatoren? ......................................................... 17
Hva er AFib? ................................................................................... 17％IHB/AFib ...................................................................................... 17
Trykkstolpeindikator ........................................................................ 18
WHO-klassifiseringsindikator .......................................................... 18
Om blodtrykk................................................................................... 18
Hva er blodtrykk? ...................................................................... 18
Hva er hypertensjon, og hvordan kan det kontrolleres? ............ 19
Hvorfor måle blodtrykket hjemme? ........................................... 19
WHOs klassifisering av blodtrykk .............................................. 19
Variasjoner i blodtrykket ............................................................ 19
Feilsøking ....................................................................................... 20
Vedlikehold ..................................................................................... 21
Tekniske spesifikasjoner ................................................................. 21
Norsk 1Kjære kunder
Gratulerer med anskaffelsen av en førsteklasses A&D blodtrykksmåler.
Enheten er utviklet for å være et brukervennlig og nøyaktig hjelpemiddel iden daglige overvåkingen av blodtrykket.
Vi anbefaler å lese denne håndboken nøye før du bruker enheten for
første gang.

Innledende merknader
Denne enheten samsvarer med EU-direktiv 93/42 EEC for medisinsk utstyr.
Samsvarsmerket
bekrefter dette. (0123: Referansenummeret til
det involverte tekniske kontrollorganet)
Enheten er designet for bruk på voksne, ikke på spedbarn eller barn.
Bruksmiljø. Enheten er ment for å betjenes av brukeren i ethjemmepleiemiljø.
Denne enheten er designet for å måle blodtrykk og puls på mennesker for
diagnoseformål.

Forholdsregler
Enheten er konstruert med presisjonskomponenter. Unngå ekstreme
temperaturer, fuktighet, direkte sollys, slag og støv.
Rengjør enheten og mansjetten med en tørr, myk klut eller en klut fuktet
med vann og et nøytralt vaskemiddel. Bruk aldri alkohol, benzen, tynnereller andre sterke kjemikalier til å rengjøre enheten eller mansjetten.
Unngå å brette mansjetten stramt og å lagre slangen stramt oppkveilet
i lange perioder. Slik behandling kan forkorte komponentenes levetid.
Oppbevares utenfor barns rekkevidde, da slangen og kabelen kan utgjøre en
reell kvelningsfare.
Unngå å vri luftslangen under måling. Det kan forårsake personskade
som følge av kontinuerlig mansjettrykk.
Verken enheten eller mansjetten er vanntett. Unngå at enheten og
mansjetten utsettes for regn, svette eller vann.
Målingen kan bli forstyrret hvis enheten brukes nær TV-apparater,
mikrobølgeovner, mobiltelefoner, røntgenutstyr eller annet utstyr med
sterke elektriske felt.
Trådløse kommunikasjonsenheter som nettverkssendere for hjemmebruk,mobiltelefoner, trådløse telefoner og basestasjoner, samt walkie-talkier kanpåvirke denne blodtrykksmåleren.
Det anbefales derfor å sørge for en avstand på minst 30 cm til slike
apparater.
Norsk 2

Kontroller at enheten er ren før den brukes igjen.
Brukt utstyr, deler og batterier skal ikke behandles som vanlig
husholdningsavfall, og må kasseres i samsvar med gjeldendebestemmelser.
Sørg for at strømadapteren enkelt kan kobles fra stikkontakten om
nødvendig når apparatet brukes med strømadapteren.
Ikke foreta inngrep i enheten. Det kan forårsake ulykker eller skade på
enheten.
For å måle blodtrykket må armen klemmes så hardt av mansjetten atblodstrømmen gjennom arterien stoppes midlertidig. Det kan føre til
smerte, nummenhet eller et midlertidig, rødt merke på armen. Dette
forekommer særlig hvis det foretas flere målinger etter hverandre.
Eventuell smerte, nummenhet eller røde merker forsvinner etter en stund.
For hyppig måling av blodtrykket kan forårsake skade som følge av atblodsirkulasjonen hindres. Pass på at bruken av enheten ikke fører til
langvarig hindring av blodsirkulasjonen hvis enheten brukes flere ganger
etter hverandre.
Hvis du har fått utført mastektomi, må du rådføre deg med lege før dubruker denne enheten.
Ikke la barn bruke enheten alene, og unngå å bruke enheten innenfor
rekkevidde for barn. Det kan forårsake ulykker eller skade.
Enheten inneholder små deler som kan medføre kvelningsfare hvis de ved
et uhell svelges av et barn.
Trekk strømadapteren ut av stikkontakten når den ikke er i bruk undermåling.
Bruk av tilbehør som ikke er beskrevet i denne håndboken kan gå ut oversikkerheten.
Batteriet kan bli varmt nok til å forårsake forbrenning hvis det kortsluttes.
La enheten tilpasses til omgivelsene før den brukes (om lag en time).
Det er ikke utført klinisk testing på nyfødte spedbarn eller gravide kvinner.
Enheten må ikke brukes på nyfødte spedbarn eller gravide kvinner.
Ikke berør batteriene, likestrømspluggen og pasienten samtidig. Det kan
føre til elektrisk støt.
Mansjetten må ikke blåses opp når den ikke er festet rundt overarmen.

Kontraindikasjoner
Følgende punkter er forholdsregler for riktig bruk av enheten.
Mansjetten må ikke festes på armen hvis annet medisinsk elektrisk utstyr
er satt på. Det kan føre til at utstyret ikke fungerer som det skal.
Personer som har alvorlig nedsatt blodomløp i armen, må rådføre seg
med lege før enheten brukes, for å unngå medisinske problemer.
Du må ikke stille diagnose på deg selv eller starte behandling på grunnlag
av måleresultatene. Rådfør deg alltid med en lege for å få evaluert
resultatene og behandlingen.
Norsk 3

Mansjetten må ikke settes på en arm med et åpent sår.
Mansjetten må ikke settes på en arm som brukes til intravenøst drypp
eller blodoverføring. Det kan forårsake personskade eller ulykker.
Ikke bruk enheten på steder der det finnes brannfarlig gass, for eksempel
anestesigass. Det kan forårsake eksplosjon.
Ikke bruk enheten i oksygenanrikede miljøer, for eksempel
oksygentrykkammer eller oksygentelt. Det kan forårsake brann eller
eksplosjon.

Norsk 4

Identifisering av deler
Display

Plugg forvekselstrømadapter

Luftplugg

Likestrømsplug
Knappen START

Luftkobling

Luftslange

Tilpasningsområde

Arterieposisjonsmerke

Indeksmerke
Mansjett

Batterirom

Batterideksel

1,5 V batterier(R6P, LR6 eller AA)

Display

Gjennomsnitt
MINNE
IHB/AFib-symbol%IHB/AFib

Systolisk trykk

Symbol for feil mansjettmontering

Symbol for bevegelsesfeil

Diastolisk trykk

Hjertesymbol
WHO-klassifiseringsindikator og
trykkstolpeindikator

Puls

Batteriindikator
Norsk 5

Symboler
Trykte symboler på enhetens etui
Symboler
SYS
DIA
PUL

IP

Funksjon/betydningStandby og slå på enheten.
Systolisk blodtrykk i mmHg
Diastolisk blodtrykk i mmHg
Puls per minutt
Veiledning for batteriinstallasjon
Likestrøm
Type BF: Enheten, mansjetten og slangen er konstruert for å gi
spesiell beskyttelse mot elektrisk støt.
Symbol for EC-direktivet for medisinsk utstyr
EU-representant
Produsent
Produksjonsdato
Internasjonalt beskyttelsessymbol
WEEE-merke

SN

Serienummer
Se bruksanvisningen/veiledningsheftet
Likestrømspluggens polaritet
Holdes tørt

Symboler som vises på displayet
SymbolerFunksjon/betydningAnbefalt tiltak
Vises mens måling pågår. Blinker når Målingen pågår. Sitt såpuls er registrert.
stille som mulig.
IHB/AFib-symbol
Vises hvis uregelmessig hjerterytme
registreres.
Kan tennes hvis enheten registrerersvært lett vibrasjon, for eksempel ved
skjelving eller risting.
Måleren kan vise feil
verdi. Foreta en ny
Vises hvis enheten registrerer atmåling.
kroppen eller armen har bevegd seg.
Sitt stille mens målingenpågår.
Norsk 6

Symboler
SymbolerFunksjon/betydningAnbefalt tiltak
Måleren kan vise feil verdi.
Vises under måling hvis mansjetten Sett på mansjetten på riktig
sitter løstmåte, og foreta en nymåling.
Registrert IHB/AFib-verdi i minnet%IHB/AFib =Antall registrerte
IHB/AFib-forekomster iminnetx100 [%]

Totalt antall

Tidligere målinger lagret i MINNET.
Gjennomsnittsdata
FULLT BATTERIBatteristrømindikatoren undermåling.
LAVT BATTERINIVÅ
Symbolet blinker når batterinivået
er lavt.
Ustabilt blodtrykk på grunn av
bevegelse under måling.

Erstatt alle batteriene med
nye batterier når symbolet
blinker.
Foreta en ny måling.
Sitt helt stille mensmålingen pågår.

Forskjellen mellom systolisk og
diastolisk verdi er mindre enn
10 mmHg.
Trykkverdien økte ikke under
oppblåsingen.

Sett på mansjetten på riktigmåte, og foreta en ny
Mansjetten sitter ikke som den skal. måling.
PUL-DISPLAYFEIL
Pulsen ble ikke registrert på riktigmåte.
Intern feil i blodtrykksmåleren

Norsk 7

Ta ut batteriene og trykk på
knappen START , og settderetter inn batteriene
igjen. Kontakt forhandleren
hvis feilen vedvarer.

Driftsmodus
1. Normal måling
Trykk på knappen START . Blodtrykket måles, og informasjonen lagres
i minnet. Enheten kan lagre de siste 60 målingene i minnet.

2. Hente frem data
Hvis ingenting vises, trykker du og holder
inne knappen START .

Trykk på og holdinne knappen
START i standby

Slipp knappen når gjennomsnittsverdienevises.

Verdienes nummer og de lagrede verdienevises automatisk i rekkefølge fra den sistemålingen.

Se flere detaljer om fremhenting av datai ”Hente frem data fra minnet“.

3. Slette alle lagrede verdier i minnet
Slå av enheten ved å holde inne knappen START
til “” vises.
Velg “” for å slette verdiene.
Verdiene slettes når symbolet
blinker.
Enheten slås av automatisk.

4. Målinger med ønsket systolisk trykk
Se informasjon om måling med ønsket systolisk trykk på side 15.

Norsk 8START

Bruke måleren
Sette inn/skifte batterier

Trinn 1

1. Fjern batteridekselet.

Trinn 3

2. Fjern de brukte batteriene, og settinn nye batterier i batterirommet
som vist. Pass på riktig polaritet(+ og -).
Bruk bare R6P, LR6 eller
AA-batterier.

Trinn 1
Trinn 2

3. Sett på batteridekselet.

FORSIKTIG
Sett batteriene inn i batterirommet som vist. Enheten fungerer ikke hvis
batteriene settes inn feil.
Når(symbol for LAVT BATTERINIVÅ) blinker på displayet, må allebatteriene erstattes med nye. Unngå å blande gamle og nye batterier. Det
kan forkorte batterienes levetid eller føre til at enheten ikke fungerer som
den skal.
Slå av enheten og vent minst to sekunder før batteriene skiftes.
Hvis symbolet(symbol for LAVT BATTERI) vises også etter atbatteriene er skiftet, foretar du en blodtrykksmåling. Det kan være atenheten registrerer de nye batteriene når målingen foretas.
(Symbolet for LAVT BATTERINIVÅ) vises ikke hvis batteriene er helt
utladet.
Batterilevetiden varierer avhengig av omgivelsestemperatur, og kan
forkortes i lave temperaturer. Fire nye LR6-batterier varer vanligvis ett år
ved måling to ganger hver dag.
Bruk bare den spesifiserte batteritypen. Batteriene som følger med
enheten er ment for testing av målerens funksjon, og kan ha begrenset
levetid.
Ta ut batteriene hvis enheten ikke skal brukes på en lang stund.
Batteriene kan lekke og føre til funksjonsfeil.

Norsk 9Bruke måleren
Koble til luftslangen

Luftkontakt

Sett luftpluggen bestemt inn
i luftkontakten.

Luftplugg

Koble til strømadapteren
Sett strømadapterpluggen inn
i likestrømskontakten.
Koble deretter strømadapteren til
en stikkontakt.

Likestrømsplugg

Plugg for vekselstrømadapter
Bruk den spesifiserte
strømadapteren. (Se side 22.)
Hold i støpselet når strømadapteren trekkes ut av stikkontakten.
Hold i adapterpluggen når strømadapteren trekkes ut av måleren.

Norsk 10

Bruke måleren
Velge riktig mansjettstørrelse
Det er viktig å bruke riktig mansjettstørrelse for å oppnå riktige målinger.
Hvis mansjetten ikke har riktig størrelse, kan målingen vise feil
blodtrykksverdi.
Armstørrelsen er trykt på hver mansjett.
Indeksmerket
og festeområdet på mansjetten viser om du bruker riktig
mansjett. (Se “Trykte symboler på mansjetten” på neste side)
Hvis indeksmerket
er utenfor området, må du kontakte den lokale
forhandleren for å få kjøpt en ny mansjett.
Mansjetten er en forbruksartikkel. Kjøp en ny hvis den blir slitt.
Armstørrelse
Anbefalt mansjettstørrelse
Katalognummer
31 cm til 45 cm
Stor mansjett for voksne
CUF-F-LA
22 cm til 42 cm
Universalmansjett
CUF-I
22 cm til 32 cm
Mansjett for voksne
CUF-F-A
Armstørrelse: Armens omkrets ved biceps.
Merk:
UA-767S-W er ikke ment for bruk med liten mansjett.

Sette på mansjetten
1. Legg mansjetten rundt overarmen,
om lag 1-2 cm over innsiden av
albuen, som vist.
Legg mansjetten rett på huden. Klærmellom mansjetten og huden kan
føre til svakt pulssignal og gi feilmåleresultat.

Unngå å brette opp ermet for stramt.
Mansjett
1 - 2 cm

Luftslange
Arterieposisjonsmerke

2. Hvis blodomløpet i overarmen
hindres av at ermet brettes for stramt,
kan det føre til unøyaktigemåleverdier.
3. Kontroller at indeksmerket
erinnenfor området for riktig tilpasning.

Norsk 11

Kontroller at
tekstilfestet sitter godt.

Indeksmerke

for riktig tilpasning

Bruke måleren
Trykte symboler på mansjetten
SymbolerFunksjon/betydningAnbefalt tiltak
Plasser merketpåarterien i overarmen ellerpå linje med ringfingerenpå innsiden av armen.

Arterieposisjonsmerke▲
REF

Indeksmerke
Katalognummer
Riktig tilpasning for mansjett forvoksne.
Trykt på mansjetten for voksne.
Bruk den store mansjetten
Over området trykt på mansjetten for for voksne i stedet forvoksne og universalmansjetten.
mansjetten for voksne eller
universalmansjetten.
Riktig tilpasning for universalmansjetten.
Trykt på universalmansjetten.
Riktig tilpasning for den store mansjetten
for voksne. Trykt på den store
mansjetten for voksne.
Under området trykt på mansjetten for
voksne og universalmansjetten.
Bruk mansjetten for
Under området trykt på den store
voksne i stedet for den
mansjetten for voksne.
store mansjetten for
voksne.
Lot-nummer

Stor mansjett for voksneOmråde for riktig tilpasning

A

L

Universalmansjett

S

W

L

A

L

Mansjett for voksneS

Norsk 12Bruke måleren
Foreta nøyaktige målinger
For mest mulig nøyaktige blodtrykksmålinger:
Sitt komfortabelt på en stol. Hvil armen på bordet. Ikke kryss bena. Holdføttene flatt på gulvet, og hold ryggen rett.
Slapp av i om lag fem til ti minutter før målingen.
Plasser midten av mansjetten på høyde med hjertet.
Sitt stille og unngå å snakke under målingen.
Unngå å måle like etter fysiske anstrengelser eller et bad. Hvil i 20 til
30 minutter før målingen foretas.
Prøv å måle blodtrykket på samme tidspunkt hver dag.

MålingDet er normalt at mansjetten føles svært stram mens målingen pågår. (Deter ingen grunn til bekymring.)

Etter målingen
Etter målingen trykker du på knappen START for å slå av enheten.
Fjern mansjetten og noter verdiene.
Merk: Enheten slås av automatisk om lag ett minutt etter målingen.
Vent minst tre minutter mellom målinger på samme person.

Norsk 13Målinger
Se “Merknader for nøyaktig måling“ på neste side før målingen foretas.

Normal måling
1.

Sett mansjetten på armen (helst venstre arm).
Sitt stille mens målingen pågår.

2.

Trykk på knappen START.
Alle displaysegmenter vises.
Null (0) vises, og blinker kort.
Displayet endres som vist i illustrasjonen til
høyre når målingen begynner. Oppblåsing
av mansjetten starter. Det er normalt at
mansjetten føles svært stram. En
trykkindikator vises på venstre side av
mansjetten mens oppblåsingen pågår.
Merk: Hvis du ønsker å stoppe oppblåsingenpå noe tidspunkt, kan du trykke på
knappen START igjen.

3.

4.

5.

På høyde med
hjertetTrykk påSTART

Alle displaysegmenter vises

Displayet viser
null
Oppblåsing
starterNår oppblåsingen er fullført, slippes luft ut
automatisk, og(hjertesymbolet)
Trykk bygges opp
blinker for å indikere at målingen pågår.
Symbolet blinker med hvert hjerteslag såsnart pulsen er registrert.
Merk: Hvis enheten ikke oppnår riktig trykk,
begynner enheten automatisk å blåse
opp mansjetten igjen.
Se “Målinger med ønsket systolisk trykk“ påneste side for å unngå at mansjetten blåses
opp på nytt.
Måling pågår
Når målingen er fullført, vises systolisk og
diastolisk trykkverdi, samt puls.
Gjenværende luft slippes ut av mansjetten,
og den tømmes helt.
Systolisk trykk
Diastolisk trykk
Trykk på knappen START for å slå av enheten.
Enheten slås av automatisk etter ett minutt uten WHO-klassifisering
Puls
betjening.

Merk: Vent minst tre minutter mellom målinger
på samme person.

Norsk 14

Gjenværende luft
slippes ut automatiskMålingerMålinger med ønsket systolisk trykk
UA-767S-W er konstruert for å registrere puls og blåse opp mansjetten til
systolisk trykk automatisk.
Bruk denne metoden hvis mansjetten blåses opp igjen flere ganger ellerhvis resultatene ikke vises selv om trykket faller til 20 mmHg eller mindre.
1.

Sett mansjetten på armen på høyde med
hjertet (helst venstre arm).

2.

Trykk på knappen START .

3.

Når tallet null blinker, trykker du på og
holder inne knappen START til et tallsom er om lag 30 til 40 mmHg høyere
enn forventet systolisk trykk vises.

4.

Når ønsket verdi nås, slipper duknappen START for å starte målingen.
Fortsett å måle blodtrykket som
beskrevet på neste side.

På høyde
med hjertet
Trykk på START
knappenSlipp knappen vedønsket systolisk
trykk
Trykk på og
hold inne START
knappen

Se forrige side for målinger

Merknader for nøyaktig måling

Sitt i en komfortabel stilling. Legg armen på et bord med håndflaten vendt
opp og mansjetten på høyde med hjertet.
Slapp av i om lag fem til ti minutter før målingen. Hvis du er
følelsesmessig opphisset eller nedtrykt, vil målingen gjenspeile dettestresset som en høyere (eller lavere) blodtrykksverdi enn normalt, og
pulsverdien vil vanligvis være høyere.
En persons blodtrykk varierer hele tiden, avhengig av hva personen gjør
og hva personen har spist. Det du drikker kan ha sterk og rask virkning på
blodtrykket.
Enhetens målinger er basert på pulsen. Hvis du har svært svak elleruregelmessig puls, kan det være vanskelig for enheten å fastslå
blodtrykket ditt.
Hvis enheten registrerer en unormal tilstand, stoppes målingen og et feilsymbol
vises. Se beskrivelse av symbolene på side 7.
Norsk 15

Enheten er kun ment for å brukes av voksne. Rådfør deg med legen din
før enheten brukes på et barn. Barn skal ikke bruke enheten uten tilsyn.
Den automatiske blodtrykksmålerens funksjon kan bli påvirket av høy
temperatur, fuktighet eller høyde over havet.

Hente frem data fra minnet
Merk: Enheten lagrer de siste 60 målingene i minnet.
1.

Hvis ingenting vises, trykker du og
holder inne knappen START for å
hente frem lagrede data.

Trykk og hold

START

knapp
Gjennomsnittlig systolisk
Gjennomsnittlig diastolisk
Gjennomsnittspuls

2.

Slipp knappen når
gjennomsnittsverdiene vises.

3.

Verdienes nummer og de lagredeverdiene vises automatisk
i rekkefølge fra den siste målingen.

Gjennomsnitt
for alle data

Systolisk

4.

Displayet slås av automatisk når alledata er vist.

Diastolisk
Puls
Nyeste data

Merk: Hvis du trykker på knappen START
under visning av data i minnet,
slås enheten av.

Systolisk
Diastolisk
Puls
Siste data (eldste)

Norsk 16

Hva er IHB/AFib-indikatoren?
Hvis måleren registrerer uregelmessig hjerterytme under målingen, vises
IHB/AFib-indikatoren på displayet sammen med måleverdiene.
Merk: Vi anbefaler at du kontakter din lege hvis
IHB/AFib-indikatoren vises ofte.

Hva er AFib?
Hjertet trekkes sammen på grunn av elektriske signaler som oppstår
i hjertet, og som sender blod gjennom kroppen. Atrieflimmer (AFib) oppstår
når det elektriske signalet i hjertets forkammer blir forvirret, og fører til
forstyrrelser i pulsintervallene. AFib kan føre til at blod blir stående stillei hjertet, noe som kan føre til at det dannes blodpropper. Dette er en årsak til
slag og hjerteanfall.

％IHB/AFib%IHB/AFib vises som frekvensen av den registrerte IHB/AFib-verdien.
IHB/AFib kan ikke bare registrere støy, som fysisk bevegelse, men ogsåuregelmessig hjerterytme. Derfor anbefaler vi at du kontakter legen dinhvis %IHB/AFib-nivået er høyt.
%IHB/AFib =Antall registrerte
IHB/AFib-forekomster i minnet

x 100 [%]

Totalt antall

Visning av %IHB/AFib: %IHB/AFib vises når gjennomsnittsverdiene vises.
(Se “2. Hente frem data“ i “Driftsmodus“)%IHB/AFib vises ikke hvis antall verdier i minnet er
seks eller lavere.
Visning av gjennomsnittsverdierNivå 0%IHB/AFib=0Nivå 1%IHB/AFib=1 - 9

Nivå 2%IHB/AFib=10 - 24

Vises ikke

Norsk 17Nivå 3%IHB/AFib=25 - 100

Trykkstolpeindikator
Indikatoren overvåker fremdriften i trykket under måling.
Trykk bygges opp
Luft slippes ut

Oppblåsing pågår

Måling pågår
Oppblåsing fullført

WHO-klassifiseringsindikator
Hvert segment i stolpeindikatoren
samsvarer med blodtrykksklassifiseringen
fra WHO (Verdens helseorganisasjon), som
er beskrevet på neste side.
WHO-klassifiseringsindikator
Alvorlig hypertensjon
Moderat hypertensjon
Mild hypertensjon
Høy normal
NormalOptimalEksempel:

Moderat hypertensjon: Indikatoren viser et segment basert på
gjeldende data, som samsvarer med
WHO-klassifiseringen.

Mild hypertensjon

Høy normal

Om blodtrykk
Hva er blodtrykk?
Blodtrykk er kraften som blodet utøver mot veggene i arteriene. Systolisk
trykk forekommer når hjertet trekker seg sammen. Diastolisk trykk
forekommer når hjertet utvider seg. Blodtrykk måles i mm kvikksølv (mmHg).
Det naturlige blodtrykket til en person representeres av det grunnleggende
trykket, som måles tidlig om morgenen mens man fortsatt hviler, og før man
spiser.
Norsk 18

Hva er hypertensjon, og hvordan kan det kontrolleres?
Hypertensjon er unormalt høyt blodtrykk i arteriene. Ubehandlet kan detforårsake mange helseproblemer, inkludert slag og hjerteanfall.
Hypertensjon kan kontrolleres ved å endre livsstil, unngå stress, samt ved
medisinering under tilsyn av en lege.
For å forebygge hypertensjon eller holde det under kontroll:
Ikke røyk
Mosjoner regelmessig
Reduser inntaket av salt og fettFå utført regelmessige
helsekontroller
Hold riktig vekt

Hvorfor måle blodtrykket hjemme?
Hvis blodtrykket måles ved en klinikk eller et legekontor, kan det føre til uro,som gir høyere verdier, 25 til 30 mmHg høyere enn det som måles hjemme.
Måling hjemme reduserer virkningen av ekstern påvirkning på
blodtrykkverdiene, er et supplement til legens verdier, og gir en mer
nøyaktig og fullstendig blodtrykkshistorikk.

WHOs klassifisering av blodtrykk

Variasjoner i blodtrykket

Referansemateriell, Journal of Hypertension 1999, vol. 17 nr.2
Hypertensjon grad 3 (alvorlig)
Diastolisk blodtrykk

Diagrammet viser
standarder for evaluering av
høyt blodtrykk uten hensyn
til alder, er utarbeidet av
Verdens helseorganisasjon(WHO).

Hypertensjon grad 2 (moderat)
Hypertensjon grad 1 (mild)
Høy-normal
NormalOptimalBlodtrykk

Blodtrykket til en persontvarierer sterkt på dags- og
Systolisk blodtrykk
sesongbasis. Det kan
variere med 30 til 50 mmHg på grunn av ulike tilstander i løpet av dagen.
Hos personer med høyt blodtrykk er variasjonene enda større. Blodtrykket
stiger normalt ved arbeid eller aktivitet, og faller til det laveste nivået mens
man sover. Vær derfor ikke for bekymret over resultatene i en enkeltmåling.
Foreta målinger på samme tidspunkt
Typiske variasjoner i løpet av
en dag (Målt hvert femte minutt)
hver dag med fremgangsmåten som er
Systolisk
beskrevet i denne håndboken for å finneditt normale blodtrykk. Regelmessige
avlesninger
girmer
omfattende
blodtrykkshistorikk. Pass på å notere
dato og klokkeslett når du noterer
Diastolisk
Soveblodtrykket. Rådfør deg med legen din
Klokkeslett
for å tolke blodtrykkverdiene dine.

Norsk 19

Feilsøking
ProblemMulig årsak
Ingenting vises Batteriene er utladet.
på displayet
selv om
enheten er slått Batteripolene venderikke riktig vei.
på.
Batterispenningen er for
lav.
Mansjetten(Symbolet for LAVTblåses ikke opp. BATTERINIVÅ) blinker.
Symbolet vises ikke hvis
batteriene er helt utladet.
Mansjetten sitter ikke
som den skal.

Anbefalt tiltak
Erstatt alle batteriene med nye
batterier.
Installer batteriene igjen med
pluss- og minuspolene som
angitt på batterirommet.
Erstatt alle batteriene med nye
batterier.

Sett mansjetten riktig på.

Pass på å sitte helt stille ogunngå å snakke undermålingen.
Enheten måler
ikke blodtrykket.
Sitt komfortabelt og stille.
Verdiene er for Mansjettens posisjon er Legg armen på et bord med
høye eller for
ikke riktig.
håndflaten vendt opp og
lave.
mansjetten på høyde med hjertet.
Hvis du har svært svak elleruregelmessig puls, kan det værevanskelig for enheten å fastslå
blodtrykket ditt.
Verdien avviker fra
verdien som er målt ved Se “Hvorfor måle blodtrykket
en klinikk eller ethjemme?“.
legekontor.
Annet
Ta ut batteriene. Sett inn
batteriene igjen på riktig måte,og foreta en ny måling.
Merk: Kontakt forhandleren hvis tiltakene som er beskrevet over ikke løser
problemet. Ikke prøv å åpne eller reparere dette produktet. Alle forsøkpå å gjøre det fører til at garantien blir ugyldig.
Du har bevegd på armen
under målingen.

Norsk 20

Vedlikehold
Enheten må ikke åpnes. Den inneholder skjøre elektriske komponenter og
en intrikat luftenhet som kan bli skadet. Hvis problemet ikke kan utbedresved å følge instruksjonene for feilsøking, kontakter du den lokale
forhandleren eller vår kundeserviceavdeling. Kundeservice hos A&D tilbyr
teknisk informasjon, reservedeler og enheter til autoriserte forhandlere.
Enheten er designet og fremstilt for å vare lenge. Det anbefales imidlertidå få enheten kontrollert annethvert år for å sikre riktig funksjon og
presisjon. Kontakt den lokale autoriserte forhandleren eller A&D for å få
utført vedlikehold.

Tekniske spesifikasjonerTypeMålemetodeMåleområde

UA-767S-W
Oscillometrisk måling
Pressure: 0 - 299 mmHg
Systolisk trykk: 60 - 279 mmHg
Diastolisk trykk: 40 - 200 mmHg
Puls:
40 - 180 hjerteslag / minuttMålepresisjon
Trykk:±3 mmHg
Puls:±5%
Strømforsyning
4 x 1,5 V batterier (R6P, LR6 eller AA) eller
strømadapter (TB-233C) (ikke inkludert)
Antall målinger
Ca. 700 ganger
LR6 (alkaliske batterier)Ca. 200 ganger
R6P (manganbatterier)
Ved trykkverdi 180 mmHg, romtemperatur 23 °C.
Klassifisering
Internt drevet ME-utstyr (forsyning med batterier) /klasse II (forsyning med strømadapter)
Kontinuerlig driftsmodus
Klinisk test
Ifølge ISO81060-2：2013
I den kliniske valideringsstudien ble K5 brukt på 85
forsøkspersoner for å fastslå diastolisk blodtrykk.
EMD
IEC 60601-1-2: 2014
Minne
Siste 60 målinger
Driftsvilkår
+10 til +40 °C / 15 til 85 % RH / 800 til 1060 hPa
Transport- / lagringsvilkår
-20 til +60 °C / 10 til 95 % RH / 700 til 1060 hPa
Mål
Ca. 140 [B] x 60 [H] x 105 [D] mm
Vekt
Ca. 245 g ekskludert batterier
Inntrengningsbeskyttelse Enhet:
IP20
Norsk 21

Anvendt delLevetid

Mansjett Type BF
Enhet:
5 år (ved bruk seks ganger daglig)
Mansjett: 2 år (ved bruk seks ganger daglig)
Strømadapter: 5 år (ved bruk seks ganger daglig)

Strømadapter (tilbehør)
Adapteren brukes til å koble enheten til en strømkilde i boligen. Kontakt den
lokale A&D-forhandleren for anskaffelse. Strømadapteren må inspiseres
eller skiftes periodisk.
TB-233C
Kontakt den lokale A&D-forhandleren foranskaffelse.
Strømadapteren må inspiseres eller skiftes
periodisk.
Trykte symboler på strømadapteren
SymbolerFunksjon/betydning
Kun for bruk innendørs
Enhet i klasse II
Termisk sikring
Sikring
Symbol for EC-direktivet
Symbol for EAC-sertifisering
Vekselstrømadapterpluggens polaritet
Tilbehør som selges separat
Mansjett
Katalognummer

Mansjettstørrelse
Armstørrelse
Stor mansjett for
CUF-F-LA
31 cm til 45 cm
voksne
CUF-I
Universalmansjett
22 cm til 42 cm
CUF-F-A
Mansjett for voksne
22 cm til 32 cm
Armstørrelse: Armens omkrets ved biceps.

Vekselstrømadapter

Katalognummer
TB-233C

Plugg (utløpstype)Type C

Merk: Spesifikasjonene kan endres uten varsel.
IP-klassifisering er innkapslingens beskyttelsesgrad i henhold til IEC
60529. Denne enheten er beskyttet mot faste gjenstander med 12 mm
diameter og større, for eksempel fingre. Denne enheten er ikke
vanntett.

Norsk 22

Sisällys
Hyvät asiakkaat ................................................................................ 2
Ennakkohuomautukset ..................................................................... 2
Varotoimet ........................................................................................ 2
Osien tunnistus ................................................................................. 5
Symbolit ............................................................................................ 6
Toimintatila ....................................................................................... 8
1. Tavallinen mittaus ................................................................... 8
2. Tietojen hakeminen ................................................................. 8
3. Kaikkien muistissa olevien tietojen poistaminen ..................... 8
4. Mittaus halutulla systolisella paineella..................................... 8
Monitorin käyttö ................................................................................ 9
Paristojen asennus/vaihto ........................................................... 9
Ilmaletkun liittäminen ................................................................. 10
Vaihtovirtasovittimen liittäminen ................................................ 10
Oikean mansettikoon valitseminen............................................ 11
Käsivarren mansetin asettaminen ............................................. 11
Tarkkojen mittausten tekeminen ............................................... 13
Mittaus....................................................................................... 13
Mittauksen jälkeen .................................................................... 13
Mittaukset ....................................................................................... 14
Tavallinen mittaus ..................................................................... 14
Mittaus halutulla systolisella paineella....................................... 15
Tarkkaa mittausta koskevia huomautuksia ............................... 15
Tietojen hakeminen muistista ......................................................... 16
Mikä on IHB/AFib-indikaattori? ....................................................... 17
Mikä on AFib? ................................................................................. 17％IHB/AFib ...................................................................................... 17
Paineen palkki-indikaattori .............................................................. 18
WHO-luokitusindikaattori ................................................................ 18
Tietoja verenpaineesta ................................................................... 18
Mikä on verenpaine? ................................................................. 18
Mitä kohonnut verenpaine tarkoittaa ja kuinka sitä hoidetaan?. 19
Miksi verenpainetta kannattaa mitata kotona? .......................... 19
WHO:n verenpaineluokitus ....................................................... 19
Verenpaineen vaihtelut ............................................................. 19
Vianmääritys ................................................................................... 20
Huolto ............................................................................................. 21
Tekniset tiedot ................................................................................ 21
Suomi 1Hyvät asiakkaat
Onnittelut! Olette hankkineet A&D:n huipputekniikkaa edustavan
verenpainemonitorin. Tämä helppokäyttöinen ja tarkka laite helpottaa
päivittäistä verenpainehoitoa.
Suosittelemme lukemaan tämän oppaan huolellisesti ennen laitteen
käyttöä ensimmäistä kertaa.

Ennakkohuomautukset
Tämä laite on lääkinnällisiä laitteita koskevan Euroopan direktiivin 93/42
ETY mukainen. Tämän osoittaa-yhdenmukaisuusmerkintä.
(0123: Numero viittaa asianomaiseen ilmoitettuun laitokseen)
Laite on tarkoitettu aikuisille, ei vastasyntyneille tai pikkulapsille tehtyihin
mittauksiin.
Käyttöympäristö. Laite on tarkoitettu ostajan omaan käyttöön
kotihoitoympäristöön.
Tämä laite on tarkoitettu ihmisten verenpaineen ja sydämen
lyöntitiheyden diagnostiseen mittaamiseen.

Varotoimet
Tämän laitteen valmistuksessa on käytetty tarkkuuskomponentteja.
Laitteen altistuminen äärimmäisille lämpötiloille, kosteudelle, suoralleauringonvalolle, iskuille tai pölylle pitää välttää.
Puhdista laite ja mansetti kuivalla, pehmeällä liinalla tai vedellä
kostutetulla liinalla ja neutraalilla pesuaineella. Älä koskaan käytä
alkoholia, bentseeniä, ohennusainetta tai muita voimakkaita kemikaaleja
laitteen tai mansetin puhdistamiseen.
Vältä mansetin taittamista tiukalle tai letkun säilyttämistä pitkiä aikoja
tiukkaan käärittynä, jottei osien käyttöikä lyhene.
Varmista, etteivät vauvat tai pikkulapset voi kuristua letkuun tai johtoon.
Älä kierrä ilmaletkua mittauksen aikana. Se voi aiheuttaa
loukkaantumisen mansetin jatkuvan paineen vuoksi.
Laite ja mansetti eivät ole vedenkestäviä. Älä anna sateen, hien tai veden
kastella laitetta tai mansettia.
Mittaus voi vääristyä, jos laitetta käytetään lähellä televisiota,
mikroaaltouunia, matkapuhelinta, röntgensäteitä tai muita laitteita, joilla on
voimakas sähkökenttä.
Langattomat viestintälaitteet, kuten kotiverkkolaitteet, matkapuhelimet,
langattomat puhelimet ja niiden tukiasemat sekä radiopuhelimet voivat
vaikuttaa verenpainemonitoriin.
Tämän vuoksi näihin laitteisiin on pidettävä vähintään 30 cm etäisyys.
Kun käytät laitetta uudelleen, varmista, että se on puhdas.
Suomi 2

Käytettyjä laitteita, osia ja paristoja ei saa käsitellä tavallisena
kotitalousjätteenä. Ne on hävitettävä voimassa olevien paikallisten
säädösten mukaan.
Kun käytössä on vaihtovirtasovitin, varmista, että vaihtovirtasovittimen voi
tarvittaessa irrottaa nopeasti pistorasiasta.
Älä tee laitteeseen muutoksia. Se voi johtaa onnettomuuksiin tai laitteen
vaurioitumiseen.
Verenpaineen mittaamiseksi mansetin pitää puristaa käsivartta tarpeeksi
voimakkaasti niin, että veren virtaus valtimoon keskeytyy väliaikaisesti.
Tämä voi aiheuttaa kipua, tunnottomuutta tai väliaikaisen punaisen jäljen
käsivarteen. Näin tapahtuu etenkin silloin, jos mittaus tehdään useamman
kerran peräkkäin. Kipu, tunnottomuus ja punaiset jäljet häviävät ajan
myötä.
Verenpaineen mittaaminen liian usein voi olla haitallista veren virtauksen
häiriintymisen vuoksi. Kun laitetta käytetään toistuvasti, tarkista, ettei sen
käyttö johda veren virtauksen pitkäaikaiseen heikentymiseen.
Jos sinulle on tehty mastektomia, kysy lääkäriltä ohjeita ennen laitteen käyttöä.
Älä anna lasten käyttää laitetta ilman valvontaa, äläkä käytä laitetta
paikassa, jossa se on pikkulasten ulottuvilla. Se voi johtaa onnettomuuksiin
tai vaurioihin.
Laite sisältää pieniä osia. Ne voivat aiheuttaa tukehtumisvaaran, mikäli
pikkulapset vahingossa nielaisevat ne.
Kytke vaihtovirtasovitin irti, jos sitä ei käytetä mittauksen aikana.
Muiden kuin tässä oppaassa mainittujen lisävarusteiden käyttö voi olla
turvallisuusriski.
Jos paristo joutuu oikosulkuun, se voi muuttua kuumaksi ja aiheuttaa
palovammoja.
Anna laitteen sopeutua ympäristöön ennen sen käyttöä (noin tunnin ajan).
Kliinisiä testauksia ei ole tehty vastasyntyneiden lasten tai raskaana olevien
naisten osalta. Älä käytä laitetta vastasyntyneiden lasten tai raskaana
olevien naisten hoitoon.
Älä koske paristoihin, DC-liittimeen ja potilaaseen samanaikaisesti. Se voi
aiheuttaa sähköiskun.
Älä täytä mansettia, jos sitä ei ole kääritty olkavarren ympärille.

Kontraindikaatiot
Seuraavaksi kuvataan varotoimet laitteen oikeaa käyttöä varten.
Älä laita mansettia käsivarteen, jossa on jo toinen sähköinen
lääkinnällinen laite. Laite ei silloin ehkä toimi kunnolla.
Terveysongelmien välttämiseksi henkilöiden, joilla on käsivarren vakava
verenkiertohäiriö, tulee kysyä lääkäriltä ohjeita ennen laitteen käyttöä.
Älä tee itse diagnoosia mittaustulosten perusteella äläkä aloita hoitoa
omatoimisesti. Pyydä aina lääkäriä arvioimaan tulokset ja hoidontarve.
Älä laita mansettia käsivarteen, jossa on parantumaton haava.
Älä laita mansettia käsivarteen, johon tehdään suonensisäistä tiputusta
Suomi 3

tai verensiirtoa. Se voi johtaa vammoihin tai onnettomuuksiin.
Älä käytä laitetta tilassa, jossa on syttyviä kaasuja, kuten
anestesiakaasuja. Se voi aiheuttaa räjähdyksen.
Älä käytä laitetta ympäristöissä, joissa on suuria happipitoisuuksia, kuten
korkeapaineisessa happikammiossa tai happiteltassa. Se voi johtaa
tulipaloon tai räjähdykseen.

Suomi 4

Osien tunnistus
Näyttö

Vaihtovirtasovittimen
pistoke

Ilmaliitospistoke

DC-liitin
START -painike
Ilmaliitäntä

Ilmaletku

Oikea sopivuusalue

Valtimon paikkamerkki
Osoitinmerkki
Käsivarren mansetti
Paristolokero
Paristolokeron kansi

1,5 V:n paristot(R6P, LR6 tai AA)

Näyttö

Keskiarvo
MUISTIIHB/AFib-symboli%IHB/AFib

Systolinen paine

Mansetin sopivuusvirheen symboli
Liikkeestä johtuvan virheen symboli

Diastolinen paine

Sydänmerkki
Sydämenlyöntitiheys

WHO:n luokitusindikaattorija paineen palkki-indikaattori
Pariston indikaattori
Suomi 5

Symbolit
Laitteen laukkuun merkityt symbolit
Symbolit
SYS
DIA
PUL

IP

Toiminta/Tarkoitus
Valmiustila ja laitteen päälle kytkentä.
Systolinen verenpaine yksikössä mmHg
Diastolinen verenpaine yksikössä mmHg
Sydämenlyönnit minuutissa
Pariston asennusohje
Tasavirta
Tyyppi BF: Laitteessa, mansetissa ja letkuissa on suojaus
sähköiskuja vastaan.
EU-direktiivin mukaisen lääkinnällisen laitteen merkintä
Edustaja EU:ssa
Valmistaja
Valmistuspäivä
Kansainvälinen suojausmerkintä
WEEE-merkintä

SN

Sarjanumero
Katso ohjekirjaa/-vihkoa
DC-liittimen napaisuus
Pidä kuivana

Näytössä näkyvät symbolit
Symbolit

Toiminta/Tarkoitus
Tulee näkyviin mittauksen aikana.
Vilkkuu, kun pulssi löytyy.

Suositeltu toimi
Mittaus on käynnissä.
Liiku mahdollisimman
vähän.

IHB/AFib-symboli
Tulee näkyviin, kun havaitaan
epäsäännöllinen sydämensyke.
Saattaa syttyä, jos havaitaan hienoista
tärinää, kuten vapinaa tai puistatusta.
Mittaustuloksen arvo voi
olla virheellinen. Yritä
Tulee näkyviin, kun havaitaan vartalon
mitata uudelleen.
tai käsivarren liike.
Pysy mittauksen aikana
liikkumatta.
Suomi 6

Symbolit
Symbolit

Toiminta/Tarkoitus
Tulee näkyviin mittauksen aikana, josmansetti on kiinnitetty löysästi

Suositeltu toimi
Mittaustuloksen arvo voi
olla virheellinen. Kiinnitä
mansetti oikein ja yritä
mitata uudelleen.

IHB/AFib-havaintojen määrä muistissa%IHB/AFib =Muistissa olevien
IHB/AFib-havaintojen määrä

x100 [%]

Määrä yhteensä

Aiemmat muistiin (MEMORY) tallennetut
mittaukset.
Keskiarvotiedot
FULL BATTERYPariston virtaindikaattori mittauksen
aikana.
LOW BATTERY
Vilkkuu, kun pariston virta on vähissä.
Epävakaa verenpaine mittauksen
aikaisen liikkumisen vuoksi.
Systolisen ja diastolisen arvon välinen
ero on enintään 10 mmHg.
Painearvo ei noussut täytön aikana.
Mansettia ei ole asetettu oikein.

Vaihda kaikki paristot
uusiin, kun merkki
vilkkuu.
Yritä mitata uudelleen.
Pysy mittauksen aikana
täysin liikkumatta.
Kiinnitä mansetti oikein
ja yritä mitata
uudelleen.

PUL DISPLAY ERROR
Pulssia ei havaita kunnolla.

Verenpainemonitorin sisäinen virhe

Suomi 7

Poista paristot, paina
START -painiketta ja
asenna sitten paristot
uudelleen. Jos virhe
näytetään yhä, otayhteys jälleenmyyjään.

Toimintatila
1. Tavallinen mittaus
Paina START -painiketta. Verenpaine mitataan ja tiedot tallennetaan muistiin.
Laite voi tallentaa muistiin 60 viimeisintä mittausta.

2. Tietojen hakeminen
Kun mitään ei näy, paina START -painiketta ja
pidä se pohjassa.

Valmiustilassa,
paina ja pidä
painettuna
START-painike

Vapauta painike, kun näkyviin tulevat
keskiarvotiedot.

Tietojen numero ja tallennetut tiedot
näytetään automaattisesti järjestyksessä
viimeisestä mittauksesta alkaen.

Tarkemmat tiedot tietojen muistista
hakemisesta löytyvät kohdasta “Tietojen
hakeminen muistista“.

3. Kaikkien muistissa olevien tietojen poistaminen
Kun laite sammutetaan, paina START -painiketta ja
pidä se painettuna, kunnes näkyviin tulee “”.
Valitse “” tietojen tyhjentämiseksi.
Tiedot on tyhjennetty, kun-merkki vilkkuu.
Laite sammuu automaattisesti.

START

4. Mittaus halutulla systolisella paineella
Katso sivulta 15 tiedot mittaamisesta halutulla systolisella paineella.

Suomi 8

Monitorin käyttö
Paristojen asennus/vaihto

Vaihe 1

1. Irrota paristolokeron kansi.

Vaihe 3

2. Poista käytetyt paristot ja asenna
uudet paristot paristolokeroon kuvan
mukaisesti. Varmista, että
napaisuudet (+ ja -) on asetettu oikein.
Käytä vain paristoja R6P, LR6 tai AA.

Vaihe 1

Vaihe 2

3. Asenna paristolokeron kansi.

HUOMAUTUS
Laita paristot paristolokeroon kuvatulla tavalla. Jos ne laitetaan väärin,laite ei toimi.
Kun(LOW BATTERY -merkki) vilkkuu näytössä, vaihda kaikki
paristot uusiin. Älä sekoita vanhoja ja uusia paristoja. Se voi lyhentää
paristojen käyttöikää tai aiheuttaa laitteeseen toimintahäiriön.
Vaihda paristot, kun laitteen sammumisesta on kulunut kaksi sekuntia tai
enemmän.
Jos(LOW BATTERY -merkki) näkyy, vaikka paristot on vaihdettu, teeverenpainemittaus. Laite voi tällöin tunnistaa uudet paristot.
(LOW BATTERY -merkki) ei tule näkyviin, kun paristot ovat tyhjät.
Paristojen käyttöikä vaihtelee ympäristön lämpötilan mukaan. Se voi ollalyhempi alhaisissa lämpötiloissa. Yleensä neljä uutta LR6-paristoa
kestävät noin yhden vuoden, kun niitä käytetään mittaukseen kahdesti
päivässä.
Käytä vain määritettyjä paristoja. Laitteen mukana toimitetut paristot ontarkoitettu monitorin toiminnan tarkastamiseen, joten niiden käyttöikä voi
olla lyhempi.
Poista paristot, jos laitetta ei tulla käyttämään pitkään aikaan.
Paristot saattavat vuotaa ja aiheuttaa laitteeseen toimintahäiriön.

Suomi 9

Monitorin käyttö
Ilmaletkun liittäminen

Ilmaliitäntä

Asenna ilmaliitospistoke tiukasti
ilmaliitäntään.

Ilmaliitospistoke

Vaihtovirtasovittimen liittäminen
Kytke vaihtovirtasovittimen pistoke
DC-liittimeen.
Liitä sitten vaihtovirtasovitin
pistorasiaan.

DC-liitin

Vaihtovirtasovittimen pistoke
Käytä määritettyä vaihtovirtasovitinta.
(Katso sivu 22.)
Kun irrotat vaihtovirtasovittimen pistorasiasta, tartu vaihtovirtasovittimen
runkoon ja vedä se ulos pistorasiasta.
Kun irrotat vaihtovirtasovittimen verenpainemonitorista, tartu
vaihtovirtasovittimen pistokkeeseen ja vedä se ulos monitorista.

Suomi 10

Monitorin käyttö
Oikean mansettikoon valitseminen
On tärkeää käyttää oikeankokoista mansettia, jotta mittaustulokset olisivat
tarkkoja. Jos mansetti ei ole oikeankokoinen, verenpainemittauksen tulos
voi olla virheellinen.
Käsivarren koko on merkitty jokaiseen mansettiin.
Mansettiin merkitty osoitin
ja oikea sopivuusalue kertovat, onko käytössä
oikea mansetti. (Katso seuraavalta sivulta “Mansettiin merkityt symbolit”.)
Jos osoitin
osuu alueen ulkopuolelle, ota yhteyttä paikalliseen
jälleenmyyjään ja hanki toinen mansetti.
Käsivarren mansetti on kuluva tuote. Jos se on kulunut, hanki tilalle uusi.
Käsivarren koko
Suositeltu mansetin koko
Kataloginumero
31 cm–45 cm
Suuri aikuisten mansetti
CUF-F-LA
22 cm–42 cm
Laajan kokoalueen mansetti
CUF-I
22 cm–32 cm
Aikuisten mansetti
CUF-F-A
Käsivarren koko: Ympärysmitta hauiksen kohdalla.
Huom.:
Mallia UA-767S-W ei ole suunniteltu pienen mansetin
käyttöön.

Käsivarren mansetin asettaminen
1. Kääri mansetti olkavarren ympärille,
noin 1–2 cm kyynärpään sisäpuolen
yläpuolelle, kuten kuvassa.
Laita mansetti suoraan iholle.
Vaatteet voivat heikentää pulssin
tuntumista ja aiheuttaa siksi
mittausvirheen.

Älä rullaa paidanhihaa kireälle
Mansetti
1 - 2 cm
Ilmaletku
Valtimon paikkamerkki

2. Jos tiukalle kääritty hiha puristaa
olkavartta, tarkkoja mittaustuloksia eiehkä saada.

Kiinnitä kangaskiinnike
tiukasti.

3. Varmista, että osoitin
osuu
oikean sopivuusalueen sisään.
Osoitin

Suomi 11

ja oikea sopivuusalue

Monitorin käyttö
Mansettiin merkityt symbolit
Symbolit

Toiminta/Tarkoitus

Suositeltu toimi
Aseta -merkki olkavarren
valtimon päälle tai käsivarren
sisäpuolella samaan linjaan
nimettömän kanssa.

Valtimon paikkamerkki▲
REF

Osoitin
Kataloginumero
Aikuisten mansetin oikea
sopivuusalue.
Merkitty aikuisten mansettiin.
Alueen yli, merkitty aikuistenmansettiin ja laajan kokoalueen
mansettiin.

Käytä suurta aikuistenmansettia aikuisten mansetin
tai laajan kokoalueen
mansetin sijaan.

Laajan kokoalueen mansetin oikea
sopivuusalue. Merkitty laajan
kokoalueen mansettiin.
Suuren aikuisten mansetin oikea
sopivuusalue. Merkitty suureen
aikuisten mansettiin.
Alueen alle, merkitty aikuistenmansettiin ja laajan kokoalueen
mansettiin.
Käytä aikuisten mansettia
suuren aikuisten mansetinsijaan.

Alueen alle, merkitty suureen
aikuisten mansettiin.
Eränumero
Suuri aikuisten mansetti

Oikea sopivuusalue

A

L

Laajan kokoalueen mansetti

S

W

L

A

L

Aikuisten mansettiS

Suomi 12

Monitorin käyttö
Tarkkojen mittausten tekeminen
Verenpainemittaus antaa tarkimmat tulokset, kun toimitaan seuraavasti:
Istu mukavasti tuoliin. Laita käsivarsi lepäämään pöydälle. Älä laita jalkoja
ristiin. Pidä jalat tasaisesti lattiassa ja suorista selkä.
Rentoudu viidestä kymmeneen minuuttia ennen mittauksen tekemistä.
Laita mansetin keskikohta sydämen tasolle.
Pysy paikallasi ja puhumatta mittauksen ajan.
Älä mittaa välittömästi fyysisen rasituksen tai kylvyn jälkeen. Lepää
20–30 minuuttia ennen mittauksen tekemistä.
Pyri mittaamaan verenpaine joka päivä samaan kellonaikaan.

Mittaus
On normaalia, että mansetti tuntuu mittauksen aikana hyvin tiukalta.
(Älä säikähdä.)

Mittauksen jälkeen
Sammuta laite mittauksen jälkeen painamalla START -painiketta.
Poista mansetti ja kirjaa tiedot muistiin.
Huom.:

Laitteessa on automaattinen sammutustoiminto, joka sammuttaa
laitteen noin minuutin kuluttua mittauksesta.
Odota vähintään kolme minuuttia ennen mittauksen tekemistä samalle
henkilölle.

Suomi 13

Mittaukset
Katso ennen mittauksien tekemistä kohta “Tarkkaa mittausta koskevia
huomautuksia“ seuraavalta sivulta.

Tavallinen mittaus
1.

Laita mansetti käsivarteen (mieluiten vasempaan).
Sydämen tasolla
Istu hiljaa mittauksen ajan.

2.

Paina START -painiketta.
Kaikki näytön alueet ovat näkyvissä.
Nolla (0) näkyy hetken vilkkuvana.
Näyttö muuttuu mittauksen alkaessa
oikeanpuoleisessa kuvassa osoitetulla
tavalla. Mansetti alkaa täyttyä. On
normaalia, että mansetti tuntuu hyvintiukalta. Täytön aikana näytön vasemmassa
reunassa näkyy paineen palkki-indikaattori.
Huom.: Täytön voi keskeyttää milloin
tahansa painamalla uudelleen
START -painiketta.

3.

4.

5.

Paina
Kaikki näytön
alueet näkyvissä

Nolla-näyttö
Täyttö aloitetaan

Kun mansetti on täytetty, se alkaa tyhjentyä
automaattisesti ja(sydänmerkki)
vilkkuu ilmoittaen, että mittaus on
käynnissä. Kun pulssi löytyy, merkki vilkkuu
jokaisella sykkeellä.
Huom.: Jos riittävää painetta ei saavuteta,mansetti alkaa täyttyä uudelleen
automaattisesti.
Jos haluat estää uudelleen täyttymisen,katso seuraavalta sivulta “Mittaus
halutulla systolisella paineella“.
Kun mittaus on valmis, systolisen ja
diastolisen paineen lukemat sekä
sydämen lyöntitiheys tulevat näyttöön.
Mansettiin jäänyt ilma poistetaan ja setyhjenee täysin.

START

Paineistus

Mittauskäynnissä

Systolinen paine
Diastolinen paine
Sammuta laite painamalla START -painiketta. WHO:n luokitus
Laite sammuu automaattisesti, jos sitä ei
Sydämen lyöntitiheys
käytetä minuutin kuluessa.

Huom.: Odota vähintään kolme minuuttia ennen
mittauksen tekemistä samalle henkilölle.
Suomi 14

Poistaa jäljelle jääneen
ilman automaattisestiMittaukset
Mittaus halutulla systolisella paineella
UA-767S-W on suunniteltu etsimään pulssi ja täyttämään mansetti
systolisen paineen tasolle automaattisesti.
Käytä tätä menetelmää, jos mansetti täyttyy toistuvasti uudelleen tai jos
tuloksia ei näytetä, vaikka paine laskee arvoon 20 mmHg tai
alhaisemmaksi.
1.

Laita mansetti käsivarteen sydämen tasolle(mieluiten vasempaan).

Sydämen tasolla

2.

Paina START -painiketta.

Paina START-painiketta

3.

Kun nolla vilkkuu, paina START painiketta ja pidä se painettuna,kunnes näkyviin tulee lukema, joka on
noin 30–40 mmHg korkeampi kuin
odotettu systolinen paine.

4.

Kun haluttu numero saavutetaan,vapauta START -painike mittauksen
aloittamiseksi. Jatka verenpaineen
mittausta edellisellä sivulla kuvatulla
tavalla.

Vapauta painike
halutun systolisen
paineen kohdalla

Paina ja pidä
painettuna
painike

Tarkkaa mittausta koskevia huomautuksia

START

Katso mittauksen tiedot
edelliseltä sivulta

Istu alas mukavaan asentoon. Laita käsivarsi pöydälle niin, että kämmen
on ylöspäin ja mansetti on sydämen tasolla.
Rentoudu viidestä kymmeneen minuuttia ennen mittauksen tekemistä.
Jos olet kiihtynyt tai henkisesti rasittunut, tämä näkyy mittauksessa:
mitattu verenpaine on normaalia korkeampi (tai matalampi) ja pulssi on
yleensä tavallista nopeampi.
Yksilön verenpaine vaihtelee jatkuvasti riippuen siitä, mitä hän tekee tai
mitä hän on syönyt. Juomilla voi olla voimakas ja nopea vaikutus
verenpaineeseen.
Laitteen tekemät mittaukset pohjautuvat sydämen sykkeeseen. Jos syke
on erittäin heikko tai epäsäännöllinen, laitteen voi olla vaikea mitata
verenpainetta.
Jos laite huomaa epänormaalit olosuhteet, se keskeyttää mittauksen ja näyttää
virhesymbolin. Katso sivulta 7 symboleiden kuvaus.
Suomi 15

Tämä laite on tarkoitettu aikuisten käyttöön. Kysy ohjeita lääkäriltä ennen
laitteen käyttöä lapsen verenpaineen mittaamiseen. Lapset eivät saa
käyttää tätä laitetta ilman valvontaa.
Liian korkeat lämpötilat tai suuri kosteus tai korkeus voivat vaikuttaa
automaattisen verenpainemonitorin toimintakykyyn.

Tietojen hakeminen muistista
Huom.: Tämä laite tallentaa muistiin 60 viimeisintä mittausta.
1.

Kun mitään ei näy, paina ja pidä
painettuna START -painiketallennettujen tietojen
hakemiseksi.

Paina ja pidä START -painike
painettuna
Systolinen keskiarvo
Diastolinen keskiarvo
Pulssin keskiarvo

Kaikkien tietojen
keskiarvo

2.

Vapauta painike, kun näkyviin
tulevat keskiarvotiedot.

3.

Tietojen numero ja tallennetut tiedot
näytetään automaattisesti
järjestyksessä viimeisestä
mittauksesta alkaen.

Systolinen
Diastolinen

4.

Näyttö sammuu automaattisesti, kun
kaikki tiedot on näytetty.

PulssiViimeisimmät tiedot

Huom.: Jos painat START -painiketta
tietojen haun aikana, laite
sammuu.

Systolinen
Diastolinen
PulssiViimeinen tieto (vanhin)
Suomi 16

Mikä on IHB/AFib-indikaattori?
Jos monitori havaitsee epäsäännöllisen rytmin mittausten aikana,
IHB/AFib-indikaattori tulee näyttöön mittausarvojen kanssa.
Huom.: Kehotamme
ottamaanyhteyttä
lääkäriin,
IHB/AFib-indikaattori näytetään toistuvasti.

jos

Mikä on AFib?
Sydän supistuu sydämessä esiintyvien sähköisten signaalien vuoksi ja selähettää verta kaikkialle kehoon. Eteisvärinää (AFib) syntyy, kun
sähköinen signaali muuttuu eteisessä epäselväksi saaden aikaan
häiriöitä pulssirytmissä. AFib voi muodostaa sydämeen verihyytymiä.
Nämä voivat helposti synnyttää verisuonitukoksia, jotka puolestaan voivat
aiheuttaa aivohalvauksen ja sydänkohtauksen.

％IHB/AFib%IHB/AFib on havaittujen IHB/AFib-havaintojen taajuus.
IHB/AFib voi havaita kohinan, esim. fyysisten liikkeiden lisäksi myös
epäsäännöllisen sydämen sykkeen. Siksi suosittelemme ottamaan
yhteyttä lääkäriin, jos %IHB/AFib-taso on korkea.
Muistissa olevien
IHB/AFib-havaintojen määrä
x 100 [%]
%IHB/AFib =Määrä yhteensä
Arvon %IHB/AFib näyttö: %IHB/AFib näytetään, kun näytetään keskiarvoja.
(Katso “2. Tietojen hakeminen“ kohdasta “Toimintatila“)%IHB/AFib-lukemaa ei näytetä, jos määrä muistissa on
kuusi tai vähemmän.
Keskiarvon näyttö

Taso 0
Taso 1
Taso 2
Taso 3%IHB/AFib=0 %IHB/AFib=1 - 9 %IHB/AFib=10 - 24 %IHB/AFib=25 - 100
Ei näytetä

Suomi 17

Paineen palkki-indikaattori
Indikaattori seuraa paineen muuttumista mittauksen aikana.
Paineistus

Ilmaa vapautetaan

Täyttö käynnissä

Mittaus käynnissä
Mansetti täynnä

WHO-luokitusindikaattori
Jokainen palkki-indikaattorin alue
vastaa WHO:n verenpaineluokitusta,
joka on kuvattu seuraavalla sivulla.
WHO:n luokitusindikaattori
Erittäin kohonnut verenpaine
Kohtalaisesti kohonnut verenpaine
Lievästi kohonnut verenpaine
Korkea normaali
Normaali
Ihanteellinen

Esimerkki:

Kohtalaisesti kohonnut verenpaine

Lievästi kohonnut verenpaine: Indikaattori näyttää, nykyisiin tietoihin
perustuen, alueen, joka vastaa
WHO:n luokitusta.
Korkea normaali

Tietoja verenpaineesta
Mikä on verenpaine?
Verenpaine on voima, jolla veri painaa valtimoiden seinämää. Systolinen
paine syntyy, kun sydän supistuu. Diastolinen paine syntyy, kun sydän
laajenee. Verenpaine mitataan elohopeamillimetreinä (mmHg). Yksilön
luonnollista verenpainetta edustaa peruspaine, joka mitataan heti aamulla,
kun henkilö on vielä levossa, ennen aamiaista.

Suomi 18

Mitä kohonnut verenpaine tarkoittaa ja kuinka sitä hoidetaan?
Kohonnut verenpaine tarkoittaa epätavallisen korkeaa valtimoverenpainetta,
joka hoitamattomana voi aiheuttaa monia terveysongelmia, mukaan lukien
aivohalvauksen tai sydänkohtauksen. Kohonnutta verenpainetta voi hoitaa
elämäntapamuutoksella, stressiä välttämällä sekä lääkkeillä lääkärin
valvonnassa.
Kohonneen verenpaineen ehkäiseminen tai hoitaminen:
Älä tupakoi
Harrasta säännöllistä liikuntaa
Vähennä suolaa ja rasvaa
Tarkista fyysinen kunto säännöllisesti
Ylläpidä hyvää painoindeksiä

Miksi verenpainetta kannattaa mitata kotona?
Verenpaineen mittaaminen sairaalassa tai lääkärin vastaanotolla voi
aiheuttaa hermostuneisuutta ja johtaa lukemiin, jotka ovat 25–30 mmHg
korkeammat kuin kotona mitatut. Kotona tehdyt mittaukset minimoivat
verenpainelukemiin vaikuttavat ulkoiset tekijät, täydentävät lääkäreiden
tekemiä mittauksia ja antavat tarkemman ja kattavamman kuvan
verenpaineen muutoksista ajan kuluessa.

WHO:n verenpaineluokitus

Verenpaineen vaihtelut

Lähde: Journal of Hypertension 1999, Vol 17 No.2
Luokan 3 kohonnut verenpaine (vakava)

Diastolinen verenpaine

Maailman terveysjärjestö(WHO) on määrittänyt
standardit korkean
verenpaineen arvioinnille,
ikään katsomatta, kaaviossa
kuvatun mukaisesti.

Luokan 2 kohonnut verenpaine (kohtalainen)

Luokan 1 kohonnut verenpaine (lievä)
Korkea-normaali
Normaali

Verenpaine

Yksilön verenpaine vaihtelee
Ihanteellinen
huomattavasti päivittäin ja
kausittain. Paine voi
Systolinen verenpaine
vaihdella 30–50 mmHg
päivän eri tilanteissa. Henkilöillä, joilla on kohonnut verenpaine, vaihtelut
voivat olla jopa suurempia. Normaalisti verenpaine kohoaa työnteon tai
leikkimisen aikana, ja se laskee alimmilleen henkilön nukkuessa. Tämän
vuoksi yhdestä mittaustuloksesta ei kannata huolestua liikaa.
Voit selvittää normaalin verenpaineesi
Tyypillinen vaihtelu päivän aikana(mitataan viiden minuutin välein)tekemällä mittauksen joka päivä samaan
Systolinen
kellon aikaan tässä oppaassa kuvattuja
menetelmiä noudattaen. Säännöllisesti
tehdyt mittaukset antavat kattavamman
kuvan verenpaineen muutoksista ajan
kuluessa.
Muistamerkitäylös
Diastolinen
Nukkuaverenpainemittauksen päivämäärä jakellonaika. Pyydä lääkäriä tulkitsemaan
Kellonaika
verenpainetietoja.
Suomi 19

Vianmääritys
Ongelma

Mahdollinen syy
Paristot ovat tyhjät.

Näytössä ei näy
mitään, vaikka Pariston navat eivät ole
laite on päällä. oikeilla paikoilla.

Suositeltu toimi
Vaihda kaikki paristot uusiin.
Asenna paristot uudelleen niin,
että negatiiviset ja positiiviset
navat on asetettu
paristolokerossa ilmoitetun
mukaisesti.

Pariston jännite on liian
alhainen.
(LOW BATTERY Mansetti ei
Vaihda kaikki paristot uusiin.
merkki) vilkkuu. Jos
täyty.
paristot ovat tyhjentyneet
täysin, merkki ei tulenäkyviin.
Mansettia ei ole asetettu
Laita mansetti oikein.
oikein.
Varmista, että potilas pysyy
Käsivartta tai vartaloa onpaikallaan ja puhumatta
liikutettu mittauksen aikana.
mittauksen ajan.
Laite ei mittaa.
Lukemat ovat
Istu mukavasti ja liikkumatta.
liian korkeita tai Mansetti on väärässä
Laita käsivarsi pöydälle niin, että
liian alhaisia.
kohdassa.
kämmen on ylöspäin ja mansetti on
sydämen tasolla.
Jos syke on erittäin heikko tai
epäsäännöllinen, laitteen voi ollavaikea mitata verenpainetta.
Arvo poikkeaa
sairaalassa tai lääkärin Katso “Miksi verenpainetta
vastaanotolla mitatusta kannattaa mitata kotona?“.
arvosta.
Muu
Poista paristot. Laita ne oikein
takaisin ja yritä mitata
uudelleen.
Huom.: Jos yllä kuvatut toiminnot eivät ratkaise ongelmaa, ota yhteyttä
jälleenmyyjään. Älä yritä avata tai korjata tuotetta; se voisi mitätöidä
takuun.

Suomi 20

HuoltoÄlä avaa laitetta. Laitteessa on hienoja sähkökomponentteja ja
monimutkainen ilmayksikkö, jotka voisivat vahingoittua. Jos ongelmaa ei
voida ratkaista vianmääritysohjeita noudattamalla, ota yhteyttä alueesi
valtuutettuun jälleenmyyjään tai asiakaspalveluosastoomme. A&D:nasiakaspalvelu toimittaa valtuutetuille jälleenmyyjille teknisiä tietoja,
varaosia ja yksiköitä.
Laite on suunniteltu ja valmistettu niin, että sen käyttöikä on pitkä. Laiteon kuitenkin hyvä tarkistaa 2 vuoden välein sen asianmukaisen toiminnan
ja tarkkuuden varmistamiseksi. Ota yhteyttä alueesi valtuutettuun
jälleenmyyjään tai A&D:hen huollon tekemiseksi.

Tekniset tiedot
Tyyppi
Mittausmenetelmä
Mittausalue

UA-767S-W
Oskillometrinen mittaus
Paine: 0 - 299 mmHg
Systolinen paine: 60 - 279 mmHg
Diastolinen paine: 40 - 200 mmHg
Pulssi: 40 - 180 sykettä/minuutti
Mittaustarkkuus
Verenpaine:±3 mmHg
Pulssi: ±5 %Virtalähde
4 x 1,5 V paristot (R6P, LR6 tai AA) tai
vaihtovirtasovitin (TB-233C) (ei sisälly)Mittausten määrä
Noin 700 kertaaLR6 (alkaliparistot)
Noin 200 kertaaR6P (mangaaniparistot)
Kun verenpaine on 180 mmHg, huoneen lämpötila
23 °C.
Luokitus
Sähköinen lääkinnällinen laite sisäisellä
virtalähteellä (paristot) / Luokka II (sovitin) Jatkuva
toimintatila
Kliininen testi
Noudattaen standardia ISO81060-2：2013
Kliinisessä validointitutkimuksessa K5:ttä käytettiin
85 koehenkilön diastolisen verenpaineen
mittaamiseen.
EMD
IEC 60601-1-2: 2014
Muisti
60 viimeistä mittausta
Toimintaolosuhteet
10 – 40°C / 15 – 85% suht. kosteus / 800 – 1060hPa
Kuljetus-/säilytysolosuhteet -20 – 60°C / 10 – 95% suht. kosteus / 700 – 1060hPa
Mitat
Noin 140 [L] x 60 [K] x 105 [S] mm
Paino
Noin 245 g, ilman paristoja
Suomi 21

Kotelointiluokka

Laite:

IP20

Liitäntäosa
Käyttöikä

Mansetti
Tyyppi BF
Laite:
5 vuotta (kun käytetään kuusi kertaa päivässä)Mansetti: 2 vuotta (kun käytetään kuusi kertaa päivässä)Vaihtovirtasovitin: 5 vuotta (kun käytetään kuusi
kertaa päivässä)Lisävaruste vaihtovirtasovitin
Sovittimella laite kytketään kodin virtalähteeseen. Sovittimen voi hankkia
ottamalla yhteyttä paikalliseen A&D-jälleenmyyjään. Vaihtovirtasovitin pitää
tarkistaa tai vaihtaa säännöllisesti.
Sovittimen voi hankkia ottamalla yhteyttä
TB-233C
paikalliseen A&D-jälleenmyyjään.
Vaihtovirtasovitin pitää tarkistaa tai vaihtaa
säännöllisesti.
Vaihtovirtasovittimeen merkityt symbolit
Symbolit
Toiminta/Tarkoitus
Vain sisäkäyttöön
Luokan II laite
Lämpösulake
SulakeEU-direktiivin mukaisen laitteen
merkintä
EAC-sertifioidun laitteen merkintä
Vaihtovirtasovittimen pistokkeen
napaisuus
Lisävarusteet myydään erikseen
Mansetti Kataloginumero
Mansetin koko
Käsivarren koko
CUF-F-LA
Suuri aikuisten mansetti
31 cm–45 cm
CUF-I
Laajan kokoalueen mansetti
22 cm–42 cm
CUF-F-A
Aikuisten mansetti
22 cm–32 cm
Käsivarren koko: Ympärysmitta hauiksen kohdalla.
Vaihtovirtasovitin

Kataloginumero
TB-233C

Pistoke (pistorasian tyyppi)
Tyyppi C

Huom.: Teknisiä tietoja voidaan muuttaa ilman ennakkoilmoitusta.
IP-luokitus on koteloiden antama suojausluokka standardin
IEC 60529 mukaisesti. Laite on suojattu sellaisia vieraita esineitä
vastaan, joiden halkaisija on 12 mm tai suurempi, esim. sormet.
Laitetta ei ole suojattu vedeltä.
Suomi 22İçindekilerDeğerli Müşterilerimiz ....................................................................... 2Başlangıç Açıklamaları ..................................................................... 2Önlemler ........................................................................................... 2Parçaların Tanımı ............................................................................. 5
Semboller.......................................................................................... 6Çalışma Modu................................................................................... 8
1. Normal Ölçüm ......................................................................... 8
2. Verileri Geri Çağırma .............................................................. 8
3. Bellekte Depolanan Tüm Verileri Silme ................................... 8
4. İstenen Sistolik Basınçla Ölçüm .............................................. 8
Monitörü Kullanma ............................................................................ 9
Pilleri Takma / Değiştirme ........................................................... 9
Hava Hortumunu Bağlama ........................................................ 10
AC Adaptörünü Bağlama .......................................................... 10Doğru Kolluk Boyutunu Seçme ................................................. 11
Kolluğu Takma .......................................................................... 11Doğru Ölçümleri Yapma ............................................................ 13Ölçüm ...................................................................................... 13Ölçümden Sonra ....................................................................... 13Ölçümler ......................................................................................... 14
Normal Ölçüm ........................................................................... 14İstenen Sistolik Basınçla Ölçüm ................................................ 15Doğru Ölçümle İlgili Notlar ........................................................ 15
Bellek Verilerini Geri Çağırma ........................................................ 16
IHB/AFib Göstergesi Nedir? ........................................................... 17
AFib Nedir? ..................................................................................... 17％IHB/AFib ...................................................................................... 17
Basınç Çubuğu Göstergesi ............................................................. 18
WHO Sınıflandırma Göstergesi ...................................................... 18
Tansiyon Hakkında ......................................................................... 18
Tansiyon Nedir? ........................................................................ 18
Hipertansiyon Nedir ve Nasıl Kontrol Edilir? ............................. 19
Tansiyon Evde Neden Ölçülmelidir? ......................................... 19
WHO Tansiyon Sınıflandırması ................................................. 19
Tansiyon Değişimleri ................................................................. 19
Sorun giderme ................................................................................ 20
Bakım.............................................................................................. 21
Teknik Veriler .................................................................................. 21
Türkçe 1Değerli Müşterilerimiz
Günümüzdeki en gelişmiş monitörlerden biri olan modern A&D tansiyon
monitörünü satın aldığınız için sizi tebrik ederiz. Kullanım kolaylığı vedoğruluk için tasarlanmış bu aygıt, günlük tansiyon yönetiminizi
kolaylaştıracaktır.
Aygıtı ilk kez kullanmadan önce bu el kitabını dikkatli bir şekilde
iyice okumanızı öneririz.

Başlangıç Açıklamaları
Bu aygıt, Tıbbi Ürünler için Avrupa Direktifi 93/42 EEC’ye uygundur. Bu,
uyumluluk işareti ile belirtilmiştir. (0123: Kapsanan yetkili
kuruluşlara referans numarası)
Aygıt, yenidoğanlarda veya bebeklerde değil, yetişkinlerde kullanım içintasarlanmıştır.
Kullanım ortamı. Aygıtı, evde sağlık bakımı yapılan bir ortamda kendibaşınıza kullanabilirsiniz.
Bu aygıt, tanılanacak kişinin tansiyonunu ve nabız hızını ölçmek içintasarlanmıştır.

Önlemler
Bu aygıt yapılırken hassas bileşenler kullanılmıştır. Aşırı sıcaklık, nem,doğrudan güneş ışığı, darbe veya tozdan kaçınılmalıdır.
Aygıtı ve kolluğu kuru, yumuşak bir bezle veya su ve yumuşak deterjanla
nemlendirilmiş bir bezle temizleyin. Aygıtı veya kolluğu temizlemek için
kesinlikle alkol, benzen, tiner veya başka sert kimyasallar kullanmayın.
Kolluğu sıkıca katlamaktan veya hortumu uzun süre sıkı bir şekilde bükülü
tutmaktan kaçının, bu tür davranışlar bileşenlerin ömrünü kısaltabilir.
Hortum ve kablonun kazayla bebekleri veya küçük çocukları boğmamasına
dikkat edin.
Ölçüm sırasında hava hortumunu bükmeyin. Bu, sürekli kolluk basıncı
nedeniyle yaralanmaya neden olabilir.
Aygıt ve kolluk suya dayanıklı değildir. Yağmur, ter ve suyun aygıtı vehortumu kirletmesine izin vermeyin.
Aygıt televizyonlara, mikrodalga fırınlara, cep telefonlarına, X-ışını veya
güçlü elektrik alanına sahip diğer aygıtlara yakın kullanılırsa ölçümler
bozulabilir.
Ev ağ aygıtları, cep telefonları, kablosuz telefonlar ve baz istasyonları,
telsizler gibi kablosuz iletişim aygıtları bu tansiyon monitörünü etkileyebilir.
Türkçe 2

Bu yüzden, bu tür aygıtlarla arada minimum 30 cm bir mesafe
korunmalıdır.
Aygıt yeniden kullanılırken, aygıtın temiz olduğunu onaylayın.
Kullanılmış ekipman, parçalar ve piller normal ev atıkları olarakdeğerlendirilmez ve uygun yerel düzenlemelere göre atılmalıdır.
AC adaptörü kullanılırken, AC adaptörünün gerektiğinde hemen elektrik
prizinden çıkarılabileceğinden emin olun.
Aygıtı değiştirmeyin. Kazalara veya aygıtın hasar görmesine neden
olabilir.
Tansiyonu ölçmek için kol, arterden kan akışını geçici olarak durduracakşekilde kolluk tarafından yeterince sıkılmalıdır. Bu kolda acıya, uyuşmaya
veya geçici kırmızı izlere neden olabilir. Bu durum özellikle ölçüm art arda
tekrarlandığında görünür. Acı, uyuşma veya kırmızı izler zamanla
kaybolur.
Tansiyonu çok sık ölçmek kan akışını etkilediğinden zarara neden olabilir.
Aygıtı tekrar tekrar kullanırken aygıtın çalışmasının kan dolaşımını uzunsüre bozmadığını kontrol edin.
Mastektomi olduysanız aygıtı kullanmadan önce doktora danışın.
Çocukların aygıtı kendi başlarına kullanmasına izin vermeyin ve aygıtı
küçük çocukların erişebileceği yerlerde kullanmayın. Kazalara veya
hasara neden olabilir.
Küçük çocukların yanlışlıkla yutması durumunda boğulma tehlikesine
neden olabilen küçük parçalar bulunur.
Ölçüm sırasında kullanılmadığında AC adaptörünü prizden çekin.
Bu kılavuzda belirtilmeyen aksesuarların kullanımı güvenliği tehlikeye
sokabilir.
Pil kısa devre yaparsa ısınabilir ve yanıklara neden olabilir.
Kullanmadan önce aygıtın çevre ortamına adapte olmasına izin verin(yaklaşık bir saat).
Yeni doğan bebeklerde ve hamile kadınlarda klinik test yapılmamıştır.
Yenidoğan bebeklerde veya hamile kadınlarda kullanmayın.
Pillere, DC jakına ve hastaya aynı anda dokunmayın. Bu, elektrikçarpmasıyla sonuçlanabilir.
Kolluğu üst kol çevresine sarmadan şişirmeyin.

KontrendikasyonlarAşağıda aygıtın uygun kullanımı ile ilgili önlemler bulunmaktadır.
Kolluğu, başka bir elektrikli tıbbi ekipmanın takılı olduğu kola takmayın.
Ekipman yanlış çalışabilir.
Kolunda ciddi dolaşım bozukluğu olan hastalar, tıbbi sorunlarlakarşılaşmamak için aygıtı kullanmadan önce bir doktora başvurmalıdır.
Ölçüm sonuçları üzerinden kendi kendinize teşhis koymayın ve tedaviyebaşlamayın. İşlemin sonuçlarının değerlendirilmesi ve tedavi için her
zaman bir doktora başvurun.
İyileşmemiş yarası olan bir kola kolluğu takmayın.
Türkçe 3İntravenöz damlama veya kan nakli alan kola kolluğu takmayın.
Yaralanmaya veya kazaya neden olabilir.
Aygıtı anestezik gazlar gibi yanıcı gazların bulunduğu yerlerde
kullanmayın. Patlamaya neden olabilir.
Aygıtı, yüksek basınçlı oksijen odası veya oksijen çadırı gibi fazlakonsantre oksijenli ortamlarda kullanmayın. Yangına veya patlamaya
neden olabilir.

Türkçe 4

Parçaların Tanımı
Ekran

Hava Bağlantı Aparatı

AC adaptörü FişiDC Jakı
START Düğmesi

Hava Soketi
Uygun
Oturtma Aralığı

Hava Hortumu
Arter Konum İşareti

Dizin İşareti
KollukPil Bölmesi

Pil Kapağı

1,5V Pil(R6P, LR6 veya AA)

Ekran

Ortalam
BELLEK
IHB/AFib sembolü

Sistolik
Basınç

%IHB/AFib
Kolluk Sıkılık Hatası Sembolü

Diyastolik
Basınç

Hareket Hatası Sembolü
Kalp İşareti
WHO Sınıflandırma Göstergesi ve
Basınç Çubuğu Göstergesi

Nabız Hızı

Pil Göstergesi
Türkçe 5Semboller
Aygıt kasası üzerine basılmış semboller
Semboller
SYS
DIA
PUL

IP

Fonksiyonu / Anlamı
Aygıtı bekleme durumuna alma ve aygıtı açma.
mmHg cinsinden sistolik tansiyon
mmHg cinsinden diyastolik tansiyon
Dakikadaki nabız
Pil takma kılavuzuDoğru akım
Tip BF: Aygıt, kolluk ve boru, elektrik çarpmalarına karşı özelkoruma sağlamak için tasarlanmıştır.
EC yönergesi tıbbi aygıt etiketi
AB temsilcisiÜreticiÜretim tarihi
Uluslararası koruma sembolü
WEEE etiketi

SN

Seri numarası
Talimat kılavuzuna / kitapçığa bakın
DC jakı polaritesi
Kuru tutun

Ekranda görünen semboller
Semboller

Fonksiyonu / AnlamıÖnerilen EylemÖlçüm sürüyor.
Ölçüm devam ederken görünür. Nabız
Mümkün olduğunca
algılandığında yanıp söner.
hareketsiz kalın.
IHB/AFib sembolü
Düzensiz kalp ritmi algılandığında
görünür.
Ürperme veya sendeleme gibi çok aztitreşim algılandığında yanabilir.
Okuma yanlış değer
verebilir. Yeniden
Bir vücut veya kol hareketiölçmeyi deneyin.
algılandığında görünür.
Ölçüm sırasında
hareketsiz kalın.
Türkçe 6

Semboller
Semboller

Fonksiyonu / Anlamı
Kolluk gevşek takıldığında ölçüm
sırasında görünürÖnerilen Eylem
Okuma yanlış değer
verebilir. Kolluğu doğru
takın ve yenidenölçmeyi deneyin.

Hafızadaki IHB/AFib algılama oranı
Hafızada algılanan%IHB/AFib =IHB/AFib sayısı

x100 [%]

Toplam sayı

BELLEKTE depolanan öncekiölçümler.
Ortalama veri
DOLU PİL
Ölçüm sırasında pil gücü göstergesi.
DÜŞÜK PİLYanıp söndüğünde pil azalmıştır.
Ölçüm sırasında hareket nedeniyle
stabil olmayan tansiyon.

İşaret yanıp söndüğünde
tüm pilleri yenileriyledeğiştirin.
Yeniden ölçmeyi
deneyin. Ölçüm
sırasında çok hareketsiz
kalın.

Sistolik ve diyastolik değerler herbirinin 10 mmHg’si içindedir.
Şişme sırasında basınç değeri artmadı. Kolluğu doğru takın ve
yeniden ölçmeyi
Kolluk düzgün takılmamıştır.
deneyin.
NABIZ GÖRÜNTÜLEME HATASI
Nabız düzgün algılanmadı.
Pilleri çıkarın ve STARTdüğmesine basın vesonra pilleri yeniden
Tansiyon monitörü dahili hatası
takın. Hata iletisi
gözükmeye devam
ediyorsa bayiyebaşvurun.

Türkçe 7Çalışma Modu
1. Normal Ölçüm
START düğmesine basın. Tansiyon ölçülür ve veri bellekte depolanır. Bu

aygıt, son 60 ölçümü bellekte tutabilir.

2. Verileri Geri Çağırma
Ekranda hiçbir şey görüntülenmediğinde
START düğmesini basılı tutun.

Aygıt bekleme
durumundayken,düğmesini basılı tutun
STARTdüğmesini

Ortalama veri görüntülenirken düğmeyi
bırakın.

Veri sayısı ve depolanan veriler son ölçümden
itibaren otomatik olarak görüntülenir.

Verileri geri çağırma hakkında ayrıntılar içinbkz. “Bellek Verilerini Geri Çağırma”.

3. Bellekte Depolanan Tüm Verileri Silme
Aygıtı kapatırken, START düğmesini “görüntülenene kadar basılı tutun.
Verileri temizlemek için “” öğesini seçin.
işareti yanıp söndüğünde veriler temizlenir.
Aygıt otomatik olarak kapanır.

”

4. İstenen Sistolik Basınçla Ölçüm
İstenen sistolik basınçla ölçüm için sayfa 15’e bakın.
Türkçe 8

START

Monitörü Kullanma
Pilleri Takma / Değiştirme

Adım 1

1. Pil kapağını çıkarın.

Adım 3

2. Kullanılmış pilleri çıkarın ve yeni
pilleri polaritelerin (+ ve -) doğru
olmasına dikkat ederek gösterildiği
gibi pil bölmesine takın.
Yalnızca R6P, LR6 veya AA pilleri
kullanın.

Adım 1
Adım 2

3. Pil kapağını takın.

DİKKAT
Pilleri gösterildiği gibi pil bölmesine takın. Yanlış takılırsa aygıt çalışmaz.
Ekranda(DÜŞÜK PİL işareti) yanıp söndüğünde, tüm pilleri
yenileriyle değiştirin. Eski ve yeni pilleri karıştırmayın. Pil ömrünü azaltabilir
veya aygıtın arızalanmasına neden olabilir.
Pilleri aygıt kapatıldıktan iki veya daha çok saniye sonra değiştirin.
Piller değiştirildikten sonra bile(DÜŞÜK PİL işareti) görünürse birtansiyon ölçümü yapın. Böylece aygıt yeni pilleri tanıyabilir.
Piller bittiğinde (DÜŞÜK PİL işareti) görünmez.
Pil ömrü ortam sıcaklığına göre değişebilir ve düşük sıcaklıklarda
kısalabilir. Genel olarak, dört adet yeni LR6 pil, her gün iki kez ölçüm içinkullanıldığında yaklaşık bir yıl dayanır.
Yalnızca belirtilen pilleri kullanın. Aygıtla sağlanan piller monitör
performansını test etmek içindir ve sınırlı ömrü vardır.
Aygıt uzun süre kullanılmayacaksa pilleri çıkarın.
Piller sızabilir ve arızaya neden olabilir.

Türkçe 9

Monitörü Kullanma
Hava soketi
r soketi

Hava Hortumunu Bağlama
Hava bağlantı aparatını hava
soketine sıkıca takın.

Hava bağlantı aparatı

AC Adaptörünü Bağlama
AC adaptörü fişini DC jakına takın.
Sonra, AC adaptörünü elektrik
prizine bağlayın.
DC jakı
Belirtilen AC adaptörünü
AC adaptörü fişikullanın. (Bkz. sayfa 22.)
AC adaptörünü elektrik prizinden çıkarırken, AC adaptörü gövdesinden
tutup çekerek prizden çıkarın.
AC adaptörü fişini tansiyon monitöründen çıkarırken, AC adaptörü fişini
tutup monitörden çıkarın.

Türkçe 10

Monitörü KullanmaDoğru Kolluk Boyutunu SeçmeDoğru kolluk boyutunu kullanmak, doğru okuma için önemlidir. Kolluk
düzgün boyutta değilse okuma yanlış tansiyon değeri ölçebilir.
Kol boyutu her kolluğun üzerine basılmıştır.
Kolluk üzerindeki dizinve uygun oturtma aralığı, doğru kolluğu takıp
takmadığınızı size belirtir. (Sonraki sayfada yer alan “Kolluk üzerinde yazılı
semboller” konusuna bakın)
Dizin,
aralığın dışını gösteriyorsa yedek kolluğu satın almak için yerelbayinize başvurun.
Kolluk, bir sarf malzemesidir. Yıpranırsa yenisini satın alın.
Kol BoyutuÖnerilen Kolluk Boyutu
Katalog Numarası
31 cm - 45 cm
Geniş Yetişkin Kolluğu
CUF-F-LA
22 cm - 42 cm
Geniş Aralıklı Kolluk
CUF-I
22 cm - 32 cm
Yetişkin Kolluğu
CUF-F-A
Kol boyutu: Pazı çevresi.
Not:
UA-767S-W modeli, küçük kollukla kullanılmak için tasarlanmamıştır.

Kolluğu Takma
1. Kolluğu üst kol çevresine, dirsek
içinin yaklaşık 1 - 2 cm üstünden
gösterildiği gibi sarın.
Kıyafetle ölçüm, belirsiz nabza ve
dolayısıyla ölçümün hatayla
sonuçlanmasına neden
olabileceğinden kolluğu, doğrudan
cildin üzerine yerleştirin.

Giysinizin kolunu çok sıkı kıvırmayın.
Kollu

Hava Hortumu
Arter konum işareti

2. Giysi kolunu sıkıca yukarı
kıvırmadan kaynaklanan üst koldaki
daralma doğru okumayı önleyebilir.
3. Dizinin
uygun oturtma aralığı
içini gösterdiğini onaylayın.

Türkçe 11

1 - 2 cmKumaş kopçayı
iyice sabitleyin.

Dizin

ve uygun oturtma aralığı

Monitörü Kullanma
Kolluk üzerine basılmış semboller
Semboller

Fonksiyonu / AnlamıÖnerilen EylemÜst kolun arterinde veya
kolun iç tarafında yüzük
parmağıyla hizalı olarakişaretini ayarlayın.

Arter Konum İşareti
▲
REF

Dizin
Katalog numarasıYetişkin kolluğu için uygun oturtma
aralığı.
Yetişkin kolluğu üzerine basılıdır.
Yetişkin kolluğu veya
Aralık üstü uyarısı, yetişkin kolluğu ve geniş aralıklı kolluk yerinegeniş aralıklı kolluk üzerine basılıdır. geniş yetişkin kolluğunu
kullanın.
Geniş aralıklı kolluk için uygun
oturtma aralığı. Geniş aralıklı kolluküzerine basılıdır.
Geniş yetişkin kolluğu için uygun
oturtma aralığı. Geniş yetişkin kolluğu
üzerine basılıdır.
Aralık altı uyarısı, yetişkin kolluğu vegeniş aralıklı kolluk üzerine basılıdır.
Geniş yetişkin kolluğu
Aralık altı uyarısı, geniş yetişkin
yerine yetişkin kolluğunu
kolluğu üzerine basılıdır.
kullanın.
Parti numarasıGeniş yetişkin kolluğu

A

Uygun oturtma aralığı

LGeniş aralıklı kolluk

S

W

L

A

L

Yetişkin kolluğu

S

Türkçe 12

Monitörü KullanmaDoğru Ölçümleri Yapma
En doğru tansiyon ölçümü için:
Bir sandalye üzerine rahat bir şekilde oturun. Kolunuzu masa üzerinde
dinlendirin. Bacak bacak üzerine atmayın. Ayaklarınızı zeminde düz tutunve sırtınızı düzleştirin.
Ölçümden önce beş ila on dakika kadar dinlenin.
Kolluğun ortasını kalbinizle aynı düzeye yerleştirin.
Ölçüm sırasında hareketsiz durun ve sessiz kalın.
Fiziki egzersizden veya banyodan hemen sonra ölçüm yapmayın. Ölçümebaşlamadan önce yirmi veya otuz dakika dinlenin.
Tansiyonunuzu her gün aynı saatte ölçmeye çalışın.

ÖlçümÖlçüm sırasında, kolluğun çok sıkı hissetmesi normaldir. (Paniklemeyin).

Ölçümden SonraÖlçümden sonra, aygıtı kapatmak için START düğmesine basın.
Kolluğu çıkarın ve verilerinizi kaydedin.
Not: Aygıtta, aygıtı ölçümden yaklaşık bir dakika sonra kapatan otomatik güç
kapatma fonksiyonu vardır.
Aynı kişinin tansiyonunu ölçerken ölçümler arasında en az üç dakika
bekleyin.

Türkçe 13ÖlçümlerÖlçüme başlamadan önce, bir sonraki sayfada yer alan “Doğru Ölçümle
İlgili Notlar” konusuna bakın.

Normal Ölçüm
1.

Kolluğu kol üzerine yerleştirin(tercihen sol kol).
Ölçüm sırasında sessizce oturun.

Kalp seviyesinde2.

Şu düğmeye basın:
Tüm ekran segmentleri görüntülenir.
Kısa bir süreliğine sıfır (0) yanıp sönen birşekilde görüntülenir.
Tüm ekran segmentleriÖlçüm başladığında sağdaki şekilde
görüntülenir
gösterildiği gibi ekran değişir. Kollukşişmeye başlar. Kolluğun çok sıkı
hissettirmesi normaldir. Şişme sırasında
ekranın sol kenarında bir basınç çubuğu
Sıfır ekranı
göstergesi görüntülenir.
Şişmeyi başlatır
Not: Şişmeyi durdurmak istediğinizde,
START düğmesine yeniden basın.

3.

Şişme tamamlandığında, sönme otomatikolarak başlar ve(kalp işareti)
Basınç veriliyorölçümün sürdüğünü göstermek için yanıp
söner. Nabız algılandığında, her nabızatışında işaret yanıp söner.
Not: Uygun bir basınç sağlanmazsa aygıt
otomatik olarak yeniden şişmeye başlar.
Yeniden şişmeyi önlemek için bir sonraki
sayfada yer alan “İstenen Sistolik
Basınçla Ölçüm” konusuna bakın.

4.

Ölçüm tamamlandığında, sistolik ve
diyastolik basınç okumaları ve nabız hızı
görüntülenir.
Kolluk, kalan havayı boşaltır ve tamamensöner.

5.

Not:

START düğmesine basın.

STARTÖlçüm sürüyor
Sistolik basınç
Diyastolik basınç

Aygıtı kapatmak için START düğmesinebasın.
WHO sınıflandırması
Hiç işlem yapılmadan geçen bir dakikadan
Nabız hızı
sonra aygıt otomatik olarak kapanır.
Aynı kişiyi ölçerken ölçümler arasında
en az üç dakika bekleyin.
Türkçe 14

Kalan havayı otomatikolarak boşaltırÖlçümlerİstenen Sistolik Basınçla Ölçüm
UA-767S-W nabzı algılamak ve kolluğu sistolik basınç düzeyine otomatikolarak şişirmek üzere tasarlanmıştır.
Bu yöntemi, tekrar tekrar yeniden şişme gerçekleştiğinde veya basıncın
20 mmHg veya daha az bir seviyeye düşmesine rağmen sonuçlar
görüntülenmediğinde kullanın.
1.

Kolluğu kol üzerine, kalp seviyesinde
yerleştirin (tercihen sol kol).

2.

START düğmesine basın.

3.

Sıfır yanıp söndüğünde, beklediğiniz
sistolik basınçtan yaklaşık
30 – 40 mmHg yüksek bir sayı
görünene kadar START düğmesini
basılı tutun.

4.

Kalp seviyesindeDüğmeye START
basınİstediğiniz sayıya erişildiğinde, ölçümübaşlatmak için START düğmesinibırakın. Önceki sayfada açıklandığı
gibi tansiyonunuzu ölçmeye devamedin.

İstenen sistolik
basınçta düğmeyi
bırakın
Düğmeyi basılı
tutun

START

Ölçüm için önceki sayfaya
bakınDoğru Ölçümle İlgili Notlar

Rahat bir konumda oturun. Avuç içiniz yukarı bakacak şekilde ve kolluğu
kalbinizle aynı düzeyde tutarak kolunuzu masa üzerine yerleştirin.
Ölçüme başlamadan önce beş ila on dakika kadar dinlenin. Duygusal
stresle heyecanlandıysanız veya üzüldüyseniz bu stres, ölçüme
yansıyarak normal tansiyon okumasından daha yüksek (veya dahadüşük) bir sonuç çıkmasına neden olabilir ve nabız okuma genellikle
normalden hızlı olur.
Tansiyonunuz yaptığınız işe ve yediğinize bağlı olarak sürekli değişir.
İçtiğiniz şeyler, tansiyonunuzu çok güçlü ve hızlı bir şekilde etkileyebilir.
Bu aygıt, ölçümlerini kalp ritmine göre yapar. Çok zayıf veya düzensiz bir
kalp ritminiz varsa aygıt, tansiyonunuzu ölçmede zorlanabilir.
Aygıt anormal bir durum algıladığında, ölçmeyi durdurur ve bir hatasembolü görüntüler. Sembollerin açıklaması için bkz. sayfa 7.
Türkçe 15

Bu aygıt, yetişkinlerin kullanması için tasarlanmıştır. Bu aygıtı çocuklar
için kullanmadan önce doktorunuza danışın. Çocuklar, bu aygıtı
gözetimsiz kullanmamalıdır.
Otomatik tansiyon monitörünün performansı, aşırı sıcaklık veya nemden
ya da rakımdan etkilenebilir.

Bellek Verilerini Geri Çağırma
Not:
1.

Bu aygıt son 60 ölçümü bellekte depolar.
Hiçbir şey görüntülenmediğinde,
depolanan verileri geri çağırmak
için _START_ düğmesini basılı
tutun.

Basılı tutun

STARTdüğmesi
Ortalama sistolik
Ortalama diyastolik
Ortalama nabız

Tüm verilerin
ortalaması

2.

Ortalama veri görüntülenirkendüğmeyi bırakın.

3.

Veri sayısı ve depolanan veriler sonölçümden itibaren otomatik olarak
görüntülenir.

4.

Tüm veriler görüntülendikten sonra
ekran otomatik olarak kapanır.

Sistolik
Diyastolik
Nabız
En yeni veriler

Not:

Verileri geri çağırırken _START_düğmesine basarsanız aygıt
kapanır.

Sistolik
Diyastolik
Nabız
Son veri (En eski)
Türkçe 16

IHB/AFib Göstergesi Nedir?
Monitör, ölçümler sırasında düzensiz ritim algıladığında IHB/AFibgöstergesi, ölçüm değerleriyle birlikte ekranda görüntülenir.
Not:

Bu
IHB/AFib göstergesini sık sık görüyorsanız doktorunuzabaşvurmanızı öneririz.

AFib Nedir?
Kalp, oluşan elektrik sinyalleri nedeniyle kasılır ve tüm gövdeye kan
gönderir. Atriyumdaki elektrik sinyali karışırsa ve nabız aralıklarında
bozukluğa neden olursa atriyal fibrilasyon (AFib) oluşur. AFib, kanın
kalpte birikmesine neden olarak felç ve kalp krizini tetikleyen kan
pıhtılarına yol açar.

％IHB/AFib%IHB/AFib algılanan IHB/AFib’in Oranı olarak görüntülenir.
IHB/AFib yalnızca fiziki hareket gibi gürültüleri değil aynı zamanda
düzensiz kalp ritmini de algılayabilir. Bu yüzden, %IHB/AFib düzeyi
yüksekse doktorunuza başvurmanızı öneririz.
Bellekteki algılanan
IHB/AFib sayısı
x 100 [%]
%IHB/AFib =Toplam sayı
%IHB/AFib ekranı: %IHB/AFib, ortalama değerlerle birlikte görüntülenir.
(Bkz. “Çalışma Modu” içindeki “2. Verileri Geri Çağırma”)
Bellek numarası altı veya daha azsa %IHB/AFib
görüntülenmez.
Ortalama değer ekranı

Düzey 0
%IHB/AFib=0

Düzey 1Düzey 2Düzey 3
%IHB/AFib=1 - 9 %IHB/AFib=10 - 24 %IHB/AFib=25 - 100

Görüntülenmez

Türkçe 17

Basınç Çubuğu Göstergesi
Gösterge ölçüm sırasında basıncın ilerleyişini izler.
Basınç veriliyor

Hava bırakılıyorŞişirme devam ediyorÖlçme devam ediyorŞişirme tamam

WHO Sınıflandırma GöstergesiÇubuk göstergesinin her bir segmenti,
sonraki sayfada tanımlanan WHO tansiyon
sınıflandırmasına karşılık gelir.
WHO Sınıflandırma Göstergesi
Ciddi hipertansiyon
Orta hipertansiyon
Hafif hipertansiyon
Yüksek normal
NormalOptimumÖrnek:

Orta hipertansiyon

Hafif hipertansiyon: Gösterge geçerli verilere göre WHOsınıflandırmasına karşılık gelen bir
segment görüntüler.
Yüksek normal

Tansiyon Hakkında
Tansiyon Nedir?
Tansiyon, arter duvarlarına kan tarafından uygulanan baskı gücüdür.
Sistolik basınç kalp kasıldığında oluşur. Diyastolik basınç kalpgenişlediğinde oluşur. Tansiyon, milimetre cıva cinsinden ölçülür (mmHg).
Kişinin doğal tansiyonu, kişi hâlâ dinlenirken ve yemek yemeden öncesabah ilk iş olarak ölçülen temel basınç olarak sunulur.

Türkçe 18

Hipertansiyon Nedir ve Nasıl Kontrol Edilir?
Hipertansiyon, anormal şekilde yüksek arter tansiyonudur, ilgilenilmezse
felç ve kalp krizi dahil birçok sağlık sorununa neden olabilir. Hipertansiyon,yaşam tarzı değiştirilerek, stresten uzak durularak ve doktor kontrolünde
ilaç tedavisiyle kontrol edilebilir.
Hipertansiyonu önlemek veya kontrol altında tutmak için:
Sigara içmeyin
Düzenli egzersiz yapın
Tuz ve yağ alımınızı azaltın
Düzenli fiziki check-up yaptırın
Uygun ağırlığınızı koruyun

Tansiyon Evde Neden Ölçülmelidir?
Klinikte veya doktorun ofisinde yapılan tansiyon ölçümü korkuya neden
olabilir ve okumalar evde ölçülenden 25 - 30 mmHg daha yüksek çıkabilir.
Evde ölçüm, tansiyon okumalarındaki dış etkileri azaltır, doktorun
okumalarını destekler ve daha doğru, tam bir tansiyon geçmişi sağlar.

WHO Tansiyon Sınıflandırması

Tansiyon Değişimleri

Referans Malzeme; Journal of Hypertension 1999, Vol 17 No.2
Derece 3 Hipertansiyon (ciddi)

Diyastolik tansiyon

Dünya Sağlık Örgütü(WHO) tarafından
belirlenmiş, çizelgede
gösterildiği gibi yaşabakılmaksızın yüksek
tansiyonu değerlendiren
standartlardır.

Derece 2 Hipertansiyon (orta)
Derece 1 Hipertansiyon (hafif)
Yüksek-normal

NormalOptimuTansiyonKişinin tansiyonu günlük ve
Sistolik tansiyon
mevsimsel olarak büyükölçüde değişir. Gün boyunca çeşitli koşullara bağlı olarak 30 ila 50 mmHg
arasında değişebilir. Tansiyonu yüksek kişilerde değişimler daha da fazlaolabilir. Tansiyon, genellikle çalışırken veya oynarken yükselir ve uyku
sırasında en düşük düzeyine düşer. Bu yüzden, tek bir ölçümün sonucuüzerine çok fazla endişelenmeyin.
Gün içinde tipik dalgalanma
Normal tansiyonunuzu öğrenmek için bu(Her beş dakikada bir ölçüm)
kılavuzda
açıklananprosedürü
Sistolikkullanarak her gün aynı saatte ölçüm
yapın. Düzenli okumalar, daha kapsamlı
tansiyon geçmişi sağlar. Tansiyonunuzu
kaydederken tarih ve saati notettiğinizden
emin
olun.
TansiyonDiyastolik
Uyku
verilerinizi yorumlamak için doktorunuza
Saatdanışın.

Türkçe 19

Sorun giderme
Sorun
Aygıt açık olsa
bile ekranda
hiçbir şey
görünmüyor.

Olası Neden
Piller bitmiştir.
Pil uçları doğru
konumda değildir.

Önerilen Eylem
Tüm pilleri yenileriyle değiştirin.
Pilleri, negatif ve pozitif uçları pilbölmesinde gösterilen şekillerleeşleşecek şekilde yeniden takın.

Pil voltajı çok düşüktür.
(DÜŞÜK PİL işareti)
Kolluk şişmiyor. yanıp sönüyor. Piller
Tüm pilleri yenileriyle değiştirin.
tamamen bittiyse işaret
görünmez.
Kolluk düzgün şekilde
Kolluğu düzgün takın.
takılmamıştır.
Ölçüm sırasındaÖlçüm sırasında hareketsiz vekolunuzu veya gövdenizi
sessiz kaldığınızdan emin olun.
hareket ettirmişsinizdir.
Aygıt ölçmüyor.
Rahat ve hareketsiz oturun.
Okumalar çok
Avuç içiniz yukarı bakacak
Kolluk konumu doğru
yüksek veya
şekilde ve kolluğu kalbinizledeğil.
çok düşük.
aynı düzeyde tutarak kolunuzu
masa üzerine yerleştirin.
Çok zayıf veya düzensiz bir kalp
ritminiz varsa, aygıt
tansiyonunuzu ölçmede
zorlanabilir.
Değer, klinikte veya
Bkz. “Tansiyon Evde Neden
doktorun ofisindeÖlçülmelidir?”.
ölçülenden farklı.
DiğerPilleri çıkarın. Yerine geri takın
ve yeniden ölçmeyi deneyin.
Not:

Yukarıda açıklanan eylemler sorunu çözmezse bayiye başvurun. Buürünü açmaya veya onarmaya çalışmayın, bunu yapmaya kalkarsanız
garantiniz geçersiz olur.

Türkçe 20

BakımAygıtı açmayın. Hassas elektrikli bileşenler ve hasar görebilecek
karmaşık bir hava ünitesi kullanır. Sorun giderme talimatlarını kullanarak
sorunu gideremezseniz, bölgenizdeki yetkili bayiye veya müşteri
hizmetleri departmanımıza başvurun. A&D müşteri hizmetleri, yetkili
bayilere teknik bilgi, yedek parça ve üniteler sağlar.
Aygıt uzun servis ömrü için tasarlanmış ve üretilmiştir. Ancak, aygıtındüzgün ve doğru çalışması için genellikle 2 yılda bir incelenmesi önerilir.
Lütfen bakım için bölgenizdeki veya A&D’deki yetkili bayiye başvurun.

Teknik Veriler
TürÖlçüm yöntemi
Ölçüm aralığı

Ölçüm doğruluğu
Güç kaynağı
Ölçüm sayısı

Sınıflandırma

Klinik test

EMD
BellekÇalışma koşulları
Nakliye / Depolamakoşulları
BoyutlarAğırlık
Giriş koruması

UA-767S-W
Osilometrik ölçüm
Basınç:0 - 299 mmHg
Sistolik basınç: 60 - 279 mmHg
Diyastolik basınç: 40 - 200 mmHg
Nabız:
40 - 180 atış / dakika
Basınç:±3 mmHg
Nabız:%±5
4 x 1,5V pil (R6P, LR6 veya AA) ya da
AC adaptörü (TB-233C) (Birlikte verilmez)
Yaklaşık 700 kez
LR6 (alkalin piller)
Yaklaşık 200 kez
R6P (manganez piller)
180 mmHg basınç değeri, 23 °C oda sıcaklığı ile.
Dahili güçle çalışan ME ekipmanı (Pillerlesağlanmıştır) / Sınıf II (Adaptörle sağlanmıştır)Sürekli çalışma modu
ISO81060-2：2013’e göre
Klinik doğrulama çalışmasında, K5 diyastolik
tansiyonun tespit edilmesi için 85 gönüllü üzerinde
kullanılmıştır.
IEC 60601-1-2: 2014
Son 60 ölçüm
+10 ila +40 °C / %15 ila 85 RH / 800 ila 1.060 hPa-20 ila +60 °C / %10 ila 95 RH / 700 ila 1.060 hPa
Yaklaşık 140 [G] x 60 [Y] x 105 [D] mm
Piller hariç yaklaşık 245 g
Aygıt:
IP20
Türkçe 21

Takılan bölüm
Kullanım ömrüKollukTür BF
Aygıt:
5 yıl (günde altı kez kullanıldığında)Kolluk:2 yıl (günde altı kez kullanıldığında)AC adaptörü: 5 yıl (günde altı kez kullanıldığında)Aksesuar AC adaptörü
Adaptör aygıtı evdeki bir güç kaynağına bağlar. Satın almak için lütfen yerel
A&D bayinize başvurun. AC adaptörünün periyodik olarak incelenmesi vedeğiştirilmesi gerekir.
TB-233C
Satın almak için lütfen yerel A&D bayinizebaşvurun.
AC adaptörünün periyodik olarak incelenmesi vedeğiştirilmesi gerekir.
AC adaptörü üzerine basılmış semboller
Semboller
Fonksiyonu / Anlamı
Yalnızca iç mekanda kullanım içindir
Sınıf II aygıtı
Termal sigorta
SigortaEC yönergesi aygıtı etiketi
EAC sertifika aygıtı etiketi
AC adaptörü fişinin polaritesi
Aksesuarlar ayrıca satılır
Katalog
KollukKolluk Boyutu
Numarası
CUF-F-LAGeniş yetişkin kolluğu
CUF-IGeniş aralıklı kolluk
CUF-F-AYetişkin kolluğu
Kol boyutu: Pazı çevresi.
AC adaptörü
Not:Katalog Numarası
TB-233C

Kol Boyutu
31 cm - 45 cm
22 cm - 42 cm
22 cm - 32 cm

Fiş (Priz türü)
Tür C

Teknik özellikler önceden haber verilmeksizin değiştirilebilir.
IP sınıflandırması, IEC 60529’a uygun kapaklar tarafından sağlanan
koruma derecesidir. Aygıt, parmaklar gibi 12 mm ve üstü çaptaki katı
yabancı nesnelere karşı korunur. Bu aygıt, suya karşı dayanıklı değildir.

Türkçe 22‫اﻟﻣﺣﺗوﯾﺎت‬‫اﻟﻌﻣﻼء اﻷﻋزاء ‪2 ................................... ................................ ................................‬‬‫ﻣﻼﺣظﺎت أوﻟﯾّﺔ ‪2 .................................... ................................ ................................‬‬‫اﻟﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ ‪2 ................................. ................................ ................................‬‬‫اﻟﺗﻌرﯾف ﺑﺄﺟزاء اﻟﺟﮭﺎز ‪4 ........................................................... ................................‬‬‫اﻟرﻣوز ‪5 ............................................. ................................ ................................‬‬‫وﺿﻊ اﻟﺗﺷﻐﯾل ‪7 ..................................... ................................ ................................‬‬‫‪ .1‬اﻟﻘﯾﺎس اﻟﻌﺎدي ‪7 ............................... ................................ ................................‬‬‫‪ .2‬ﻛﯾﻔﯾﺔ ﻣراﺟﻌﺔ ﺑﯾﺎﻧﺎت اﻟذاﻛرة ‪7 .............................................. ................................‬‬‫‪ .3‬ﺣذف ﺟﻣﯾﻊ اﻟﺑﯾﺎﻧﺎت اﻟﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة ‪7 ................................. ................................‬‬‫‪ .4‬إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ ‪7 ..............................................‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس ‪8 ............................................................. ................................‬‬‫ﺗرﻛﯾب ‪ /‬ﺗﻐﯾﯾر اﻟﺑطﺎرﯾﺎت ‪8 ..................................................... ................................‬‬‫ﺗوﺻﯾل ﺧرطوم اﻟﮭواء ‪9 ......................................................... ................................‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ‪9 .................................................. ................................‬‬‫ﺗوﺻﯾل ّ‬‫اﺧﺗﯾﺎر ﻣﻘﺎس رﺑﺎط اﻟذراع اﻟﺻﺣﯾﺢ ‪10 ......................................... ................................‬‬‫ﻛﯾﻔﯾﺔ وﺿﻊ رﺑﺎط اﻟذراع ‪10 ..................................................... ................................‬‬‫ﻛﯾﻔﯾﺔ إﺟراء ﻋﻣﻠﯾﺎت ﻗﯾﺎس دﻗﯾﻘﺔ ‪12 ............................................. ................................‬‬‫اﻟﻘﯾﺎس ‪12 ......................................... ................................ ................................‬‬‫ﺑﻌد اﻟﻘﯾﺎس ‪12 .................................... ................................ ................................‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ‪13 .................................. ................................ ................................‬‬‫اﻟﻘﯾﺎس اﻟﻌﺎدي ‪13 ................................. ................................ ................................‬‬‫إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ ‪14 ................................................‬‬‫ﻣﻼﺣظﺎت ﺑﺎﻟﻧﺳﺑﺔ ﻟﻠﻘﯾﺎس اﻟدﻗﯾﻖ ‪14 ............................................. ................................‬‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة ‪15 .................................................... ................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣؤﺷر ‪) IHB/AFib‬ﻋدم اﻧﺗظﺎم ﺿرﺑﺎت اﻟﻘﻠب‪/‬اﻟرﺟﻔﺎن اﻷذﯾﻧﻲ(؟ ‪16 ..........................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪(AFib‬؟ ‪16 ...................................... ................................‬‬‫‪16 ................................. ................................ ................................ %IHB/AFib‬‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط ‪17 ............................................................ ................................‬‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ‪17 ......................................... ................................‬‬‫ﻧُﺑذة ﻋن ﺿﻐط اﻟدم ‪17 .............................................................. ................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺿﻐط اﻟدم؟ ‪17 .................................................... ................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎرﺗﻔﺎع ﺿﻐط اﻟدم وﻛﯾف ﯾﻣﻛن اﻟﺗﺣﻛم ﻓﯾﮫ؟ ‪18 ..................................................‬‬‫ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟ ‪18 ....................................... ................................‬‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم ‪18 .......................................................‬‬‫ﺣﺎﻻت اﻻﺧﺗﻼف ﻓﻲ ﻣﺳﺗوى ﺿﻐط اﻟدم ‪18 .................................... ................................‬‬‫اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ ‪19 .................................................. ................................‬‬‫اﻟﺻﯾﺎﻧﺔ ‪20 .......................................... ................................ ................................‬‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻔﻧﯾﺔ ‪20 .................................... ................................ ................................‬‬‫اﻟﻌرﺑﯾﺔ ‪1‬‬‫اﻟﻌﻣﻼء اﻷﻋزاء‬‫ﺗطورا ً ﻓﻲ‬‫ﺗﮭﺎﻧﯾﻧﺎ ﻟﻛم ﻋﻠﻰ ﺷراء ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﻣﺗﻣﯾّز ﻣن ‪ ،A&D‬أﺣد اﻷﺟﮭزة اﻷﻛﺛر ّ‬‫ﺻ ّﻣم اﻟﺟﮭﺎز ﻟﯾوﻓر ﺳﮭوﻟﺔ ﻓﻲ اﻻﺳﺗﺧدام ودﻗﺔً ﻓﻲ اﻟﻘﯾﺎس‪ ،‬ﺣﯾث ﯾﺳ ّﮭل ﻋﻠﯾك اﻟﻘﯾﺎس اﻟﯾوﻣﻲ‬‫اﻟوﻗت اﻟﺣﺎﻟﻲ‪ُ .‬‬‫ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﻧُوﺻﯾك ﺑﻘراءة ھذا اﻟدﻟﯾل ﺑﻌﻧﺎﯾﺔ ﻗﺑل اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻠﻣرة اﻷوﻟﻰ‪.‬‬‫ﻣﻼﺣظﺎت أوﻟﯾّﺔ‬‫ﯾﺗطﺎﺑﻖ ھذا اﻟﺟﮭﺎز ﻣﻊ اﻟﻣواﺻﻔﺎت اﻷوروﺑﯾﺔ ‪ 93/42 EEC‬اﻟﺧﺎﺻﺔ ﺑﺎﻷﺟﮭزة اﻟطﺑﯾﺔ‪ .‬وﺗﺷﯾر ﻋﻼﻣﺔ‬‫ﻋﻠﻰ دﻗﺔ اﻟﺗطﺎﺑﻖ‪ :0123) .‬اﻟرﻗم اﻟﻣرﺟﻌﻲ ﻟﻠﮭﯾﺋﺔ اﻟﻣﻌﻧﯾﺔ(‬‫ﺻﻣم اﻟﺟﮭﺎز ﻻﺳﺗﺧداﻣﮫ ﻣن ﻗﺑل اﻟﺑﺎﻟﻐﯾن ﻓﻘط‪ ،‬وﻟﯾس ﻋﻠﻰ اﻟرﺿّﻊ وﺣدﯾﺛﻲ اﻟوﻻدة‪.‬‬‫ُ‬‫ﻣﻛﺎن اﻻﺳﺗﺧدام‪ :‬ھذا اﻟﺟﮭﺎز ُﻣﺧﺻص ﻟﻼﺳﺗﻌﻣﺎل ﻷﻏراض اﻟرﻋﺎﯾﺔ اﻟﺻﺣﯾﺔ اﻟﻣﻧزﻟﯾﺔ‪.‬‬‫ﺻﻣم ھذا اﻟﺟﮭﺎز ﻟﻘﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم وﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟﻺﻧﺳﺎن ﻷﻏراض اﻟﺗﺷﺧﯾص‪.‬‬‫ُ‬‫اﻟﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ‬‫ﻛوﻧﺎت دﻗﯾﻘﺔ؛ ﻟذا ﺗﺟ ّﻧب ﻣطﻠ ًﻘﺎ ﺗﻌرﯾﺿﮫ ﻟدرﺟﺎت اﻟﺣرارة واﻟرطوﺑﺔ اﻟﻣرﺗﻔﻌﺔ وأﺷﻌﺔ‬‫ُ‬‫ﺻ ّﻧﻊ ھذا اﻟﺟﮭﺎز ﻣن ُﻣ ّ‬‫اﻟﺷﻣس اﻟﻣﺑﺎﺷرة واﻟﺻدﻣﺎت واﻟﻐﺑﺎر‪.‬‬‫ّ‬‫ﻧظف اﻟﺟﮭﺎز واﻟرﺑﺎط ﺑﺎﺳﺗﺧدام ﻗطﻌﺔ ﻗﻣﺎش ﺟﺎﻓﺔ وﻧﺎﻋﻣﺔ أو ﻗطﻌﺔ ﻗﻣﺎش ﻣﺑﻠﻠﺔ ﺑﺎﻟﻣﺎء وﻣﻧظف ﻣﻌﺗدل‪.‬‬‫ﺗﺟﻧب ﻣطﻠﻘًﺎ اﺳﺗﺧدام اﻟﻛﺣول أو اﻟﺑﻧزﯾن أو اﻟﺛﻧر أو ﻏﯾرھﺎ ﻣن اﻟﻣواد اﻟﻛﯾﻣﯾﺎﺋﯾﺔ اﻟﻘﺎﺳﯾﺔ ﻣن أﺟل ﺗﻧظﯾف‬‫اﻟﺟﮭﺎز أو اﻟرﺑﺎط‪.‬‬‫ﻲ اﻟرﺑﺎط ﺑﺷدة أو ﺗﺧزﯾن اﻟﺧرطوم ﺑﺛﻧﯾﮫ ﺑﺷدة ﻟﻔﺗرات طوﯾﻠﺔ؛ ﯾؤدي ھذا اﻟﺗﺻرف إﻟﻰ ﻗﺻر ﻣدة‬‫ﺗﺟﻧب ط ّ‬‫اﻟﻣﻛوﻧﺎت‪.‬‬‫اﺳﺗﺧدام ﺗﻠك‬‫ّ‬‫ﯾﺟب اﺗﺧﺎذ اﻟﺣﯾطﺔ واﻟﺣذر ﻟﻛﻲ ﻻ ﯾﺗﺳﺑب اﻟﺧرطوم واﻟﻛﺎﺑل ﻓﻲ ﺧﻧﻖ اﻷطﻔﺎل ﻋن طرﯾﻖ اﻟﺧطﺄ‪.‬‬‫ﻻ ﺗﻘم ﺑﺛﻧﻲ ﺧرطوم اﻟﮭواء أﺛﻧﺎء إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻟﺗﻌرض ﻹﺻﺎﺑﺔ ﻧظرا ً‬‫ﻟﻠﺿﻐط اﻟﻣﺳﺗﻣر ﻓﻲ اﻟرﺑﺎط‪.‬‬‫ﺗﻌرض اﻟﺟﮭﺎز واﻟرﺑﺎط ﻟﻠﺑﻠل ﺑﻔﻌل اﻷﻣطﺎر واﻟﻌرق واﻟﻣﯾﺎه‪.‬‬‫اﻟﺟﮭﺎز واﻟرﺑﺎط ﻏﯾر ﻣﻘﺎوﻣﯾن ﻟﻠﻣﺎء‪ .‬ﺗﺟﻧﱠب ّ‬‫ﯾﻣﻛن أن ﺗﺗﻌرض ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻠﺗﺷوﯾش إذا ﺟرى اﺳﺗﺧدام اﻟﺟﮭﺎز ﺑﺟوار أﺟﮭزة اﻟﺗﻠﻔزﯾون أو‬‫اﻟﻣﯾﻛرووﯾف أو اﻟﮭواﺗف اﻟﻣﺣﻣوﻟﺔ أو اﻷﺟﮭزة اﻟﺗﻲ ﯾﻧﺑﻌث ﻣﻧﮭﺎ اﻷﺷﻌﺔ اﻟﺳﯾﻧﯾﺔ أو اﻷﺧرى ذات اﻟﻣﺟﺎﻻت‬‫اﻟﻛﮭرﺑﺎﺋﯾﺔ اﻟﻘوﯾﺔ‪.‬‬‫أﺟﮭزة اﻻﺗﺻﺎﻻت اﻟﻼﺳﻠﻛﯾﺔ‪ ،‬ﻣﺛل أﺟﮭزة اﻻﺗﺻﺎل اﻟﺷﺑﻛﻲ اﻟﻣﻧزﻟﯾﺔ‪ ،‬واﻟﮭواﺗف اﻟﻣﺣﻣوﻟﺔ‪ ،‬واﻟﮭواﺗف اﻟﻼﺳﻠﻛﯾﺔ‬‫واﻟﻣﺣطﺎت اﻟﻘﺎﻋدﯾﺔ ﻟﮭﺎ‪ ،‬واﻷﺟﮭزة اﻟﻼﺳﻠﻛﯾﺔ اﻟﺗﻲ ﺗﻌﻣل ﺑﻣوﺟﺎت اﻟرادﯾو‪ ،‬ﺟﻣﯾﻌﮭﺎ ﯾﻣﻛن أن ﺗؤﺛر ﻋﻠﻰ ﺟﮭﺎز‬‫ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫وﻟذﻟك‪ ،‬ﯾﺟب اﻟﺣﻔﺎظ ﻋﻠﻰ ﻣﺳﺎﻓﺔ ﻻ ﺗﻘل ﻋن ‪ 30‬ﺳم ﺑﯾن اﻟﺟﮭﺎز وﺗﻠك اﻷﺟﮭزة‪.‬‬‫ﻋﻧد إﻋﺎدة اﺳﺗﺧدام اﻟﺟﮭﺎز‪ ،‬ﺗﺄﻛد ﻣن أﻧﮫ ﻧظﯾف‪.‬‬‫اﻷﺟﮭزة واﻷﺟزاء واﻟﺑطﺎرﯾﺎت اﻟﻣﺳﺗﻌﻣﻠﺔ ﻻ ﺗُﻌﺎﻣل ﻣﻌﺎﻣﻠﺔ اﻟﻧﻔﺎﯾﺎت اﻟﻣﻧزﻟﯾﺔ اﻟﻌﺎدﯾﺔ‪ ،‬وﯾﺟب اﻟﺗﺧﻠص ﻣﻧﮭﺎ‬‫وﻓﻘًﺎ ﻟﻠواﺋﺢ اﻟﻣﺣﻠﯾﺔ اﻟﻣﻌﻣول ﺑﮭﺎ‪.‬‬‫اﻟﻣﺣول ﻣن اﻟﻣﺄﺧذ اﻟﻛﮭرﺑﺎﺋﻲ ﻋﻧد‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪ ،‬ﺗﺄﻛد ﻣن ﺳﮭوﻟﺔ إﻣﻛﺎﻧﯾﺔ إزاﻟﺔ‬‫ﻋﻧد اﺳﺗﺧدام‬‫ّ‬‫ّ‬‫اﻟﺿرورة‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪2‬‬‫ﻻ ﺗﻌدّل ﻓﻲ اﻟﺟﮭﺎز‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ وﻗوع ﺣوادث أو ﺗﻠف اﻟﺟﮭﺎز‪.‬‬‫ﯾﺗطﻠب ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬اﻟﺿﻐط ﻋﻠﻰ اﻟذراع ﻣن ﺧﻼل اﻟرﺑﺎط ﺑﻘوة ﺗﻛﻔﻲ ﻹﯾﻘﺎف ﺗدﻓﻖ اﻟدم ﻣؤﻗﺗًﺎ‬‫ﻓﻲ ﺷرﯾﺎن اﻟذراع‪ .‬ﯾﻣﻛن أن ﯾُﺳﺑّب ذﻟك ﻓﻲ اﻟﺷﻌور ﺑﺄﻟم أو ﺗﻧﻣﯾل أو ظﮭور ﻋﻼﻣﺔ ﺣﻣراء ﻣؤﻗﺗﺔ ﺑﺎﻟذراع‪.‬‬‫ﺗظﮭر ھذه اﻟﻌﻼﻣﺎت ﻻ ﺳﯾﻣﺎ ﻋﻧد ﺗﻛرار ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس ﺑﺷﻛل ﻣﺗﺗﺎﺑﻊ‪ .‬ﺳﯾﺧﺗﻔﻲ أي أﻟم أو ﺗﻧﻣﯾل أو ﻋﻼﻣﺎت‬‫ﺣﻣراء ﺑﻣرور اﻟوﻗت‪.‬‬‫ﯾﻣﻛن أن ﯾﺗﺳﺑب ﺗﻛرار ﻋﻣﻠﯾﺎت ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﺑﺷﻛل ﻣﺗﺗﺎﺑﻊ‪ ،‬ﻓﻲ ﺣدوث ﻣﺷﻛﻠﺔ ﺻﺣﯾﺔ ﺑﺳﺑب ﺗﻛرار‬‫اﻟﺿﻐط ﻋﻠﻰ ﺗدﻓﻖ اﻟدم‪ .‬ﺗﺣﻘﻖ ﻣن أن ھذا اﻟﺗﻛرار ﻻ ﯾؤدي إﻟﻰ أﺿرار طوﯾﻠﺔ اﻷﻣد ﺑﺎﻟدورة اﻟدﻣوﯾﺔ‪.‬‬‫إذا ﻛﻧت ﻗد أﺟرﯾت ﻋﻣﻠﯾﺔ اﺳﺗﺋﺻﺎل اﻟﺛدي‪ ،‬ﻓﺎﺳﺗﺷﯾري اﻟطﺑﯾب ﻗﺑل اﺳﺗﺧدام اﻟﺟﮭﺎز‪.‬‬‫ﻻ ﺗﺳﻣﺢ ﻟﻸطﻔﺎل ﺑﺎﺳﺗﺧدام اﻟﺟﮭﺎز ﻣن ﺗﻠﻘﺎء أﻧﻔﺳﮭم وﺣﺎﻓظ ﻋﻠﻰ اﺳﺗﺧدام اﻟﺟﮭﺎز ﻓﻲ ﻣﻛﺎن ﺑﻌﯾد ﻋن ﻣﺗﻧﺎول‬‫اﻷطﻔﺎل‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ وﻗوع ﺣوادث أو ﺗﻠف‪.‬‬‫ﺛﻣﺔ أﺟزاء ﺻﻐﯾرة ﻓﻲ اﻟﺟﮭﺎز ﯾﻣﻛن أن ﺗﺳﺑب اﻻﺧﺗﻧﺎق ﻟﻸطﻔﺎل ﻓﻲ ﺣﺎل اﺑﺗﻼﻋﮭﺎ ﻋن طرﯾﻖ اﻟﺧطﺄ‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻋﻧد ﻋدم اﺳﺗﺧداﻣﮫ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﻓﺻل ّ‬‫ﯾﻌرض اﺳﺗﺧدام ﻣﻠﺣﻘﺎت ﻻ ﺗرد ﺗﻔﺻﯾ ً‬‫ﻼ ﻓﻲ ھذا اﻟدﻟﯾل ﺳﻼﻣﺗك ﻟﻠﺧطر‪.‬‬‫ﻗد ّ‬‫إذا ﺗﺳﺑﺑت اﻟﺑطﺎرﯾﺔ ﻓﻲ ﺣدوث داﺋرة ﻗﺻر‪ ،‬ﻓﯾﻣﻛن أن ﺗﺻﺑﺢ ﺳﺎﺧﻧﺔ وﺗﺗﺳﺑب ﻓﻲ ﺣروق ﺑﺎﻟﺟﻠد‪.‬‬‫اﺗرك اﻟﺟﮭﺎز ﯾﺗﻛﯾّف ﻋﻠﻰ اﻟﺑﯾﺋﺔ اﻟﻣﺣﯾطﺔ ﻗﺑل اﻻﺳﺗﺧدام )ﻣدة ﺳﺎﻋﺔ واﺣدة ﺗﻘرﯾﺑًﺎ(‪.‬‬‫ﻟم ﺗُﺟر اﺧﺗﺑﺎرات ﺳرﯾرﯾﺔ ﻋﻠﻰ اﻷطﻔﺎل ﺣدﯾﺛﻲ اﻟوﻻدة واﻟﻧﺳﺎء اﻟﺣواﻣل‪ .‬ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻋﻠﻰ اﻷطﻔﺎل‬‫ﺣدﯾﺛﻲ اﻟوﻻدة أو اﻟﻧﺳﺎء اﻟﺣواﻣل‪.‬‬‫اﻟﺗﻌرض‬‫ﻻ ﺗﻠﻣس اﻟﺑطﺎرﯾﺎت وﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر واﻟﻣرﯾض ﺳوﯾًﺎ ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ‪ .‬ﻗد ﯾؤدي ذﻟك إﻟﻰ‬‫ّ‬‫ﻟﺻدﻣﺔ ﻛﮭرﺑﺎﺋﯾﺔ‪.‬‬‫ﻻ ﺗﺑدأ ﻓﻲ ﻧ ْﻔﺦ اﻟﮭواء دون ﻟف اﻟرﺑﺎط ﺣول اﻟﻌﺿد‪.‬‬‫ﻣواﻧﻊ اﻻﺳﺗﻌﻣﺎل‬‫ﻓﯾﻣﺎ ﯾﻠﻲ اﺳﺗﻌراض ﻟﻠﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ ﻣن أﺟل اﺳﺗﺧدام ﻣﻼﺋم ﻟﻠﺟﮭﺎز‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﯾﺗﺻل ﺑﮫ ﺟﮭﺎز طﺑﻲ ﻛﮭرﺑﺎﺋﻲ آﺧر‪ .‬ﻗد ﻻ ﯾﻌﻣل اﻟﺟﮭﺎز ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﯾﺟب ﻋﻠﻰ اﻷﺷﺧﺎص اﻟذﯾن ﯾﻌﺎﻧون ﻣن ﻧﻘص ﺷدﯾد ﻓﻲ اﻟدورة اﻟدﻣوﯾﺔ ﻓﻲ اﻟذراع اﺳﺗﺷﺎرة اﻟطﺑﯾب ﻗﺑل‬‫اﻟﺗﻌرض ﻟﻠﻣﺷﻛﻼت اﻟطﺑﯾﺔ‪.‬‬‫اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﺗﻔﺎدي‬‫ّ‬‫ﻻ ﺗﺷ ّﺧص ﺑﻧﻔﺳك ﻧﺗﺎﺋﺞ اﻟﻘﯾﺎس وﺗﺑدأ ﻓﻲ ﺗﻧﺎول اﻟﻌﻼج ﻣن ﺗﻠﻘﺎء ﻧﻔﺳك‪ .‬اﺳﺗﺷر اﻟطﺑﯾب داﺋ ًﻣﺎ ﻣن أﺟل ﺗﻘﯾﯾم‬‫اﻟﻧﺗﺎﺋﺞ واﻟﻌﻼج‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﺑﮫ ﺟرح ﻟم ﯾﻠﺗﺋم ﺑﻌد‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﯾﺗﻠﻘﻰ اﻟﺗﻧﻘﯾط اﻟورﯾدي أو ﻧﻘل اﻟدم‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻟﺗﻌرض‬‫ﻟﻺﺻﺎﺑﺔ أو اﻟﺿرر‪.‬‬‫ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻓﻲ أﻣﺎﻛن وﺟود اﻟﻐﺎزات اﻟﻘﺎﺑﻠﺔ ﻟﻼﺷﺗﻌﺎل‪ ،‬ﻣﺛل ﻏﺎزات اﻟﺗﺧدﯾر‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك‬‫ﻓﻲ ﺣدوث اﻧﻔﺟﺎر‪.‬‬‫ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻓﻲ أﻣﺎﻛن ﺗرﻛﯾز اﻷوﻛﺳﺟﯾن ﻓﯾﮭﺎ ﻣرﺗﻔﻊ‪ ،‬ﻣﺛل ﻏرﻓﺔ اﻷﻛﺳﺟﯾن ﻋﺎﻟﯾﺔ اﻟﺿﻐط أو ﺧﯾﻣﺔ‬‫اﻷﻛﺳﺟﯾن‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻧدﻻع ﺣرﯾﻖ أو ﺣدوث اﻧﻔﺟﺎر‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪3‬‬‫اﻟﺗﻌرﯾف ﺑﺄﺟزاء اﻟﺟﮭﺎز‬‫اﻟﺷﺎﺷﺔ‬‫ﺻل اﻟﮭواء‬‫ﻗﺎﺑس ﻣو ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻗﺎﺑس ّ‬‫ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫اﻟزر ‪START‬‬‫ﻣﻘﺑس اﻟﮭواء‬‫ﺧرطوم اﻟﮭواء‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫ﻋﻼﻣﺔ ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫ﻋﻼﻣﺔ اﻟﻣؤﺷر‬‫رﺑﺎط اﻟذراع‬‫ﺣﺟرة اﻟﺑطﺎرﯾﺎت‬‫ﻏطﺎء اﻟﺑطﺎرﯾﺎت‬‫ﺑطﺎرﯾﺎت ﺑﺟﮭد ‪ 1.5‬ﻓوﻟت‬‫)ﻣﻘﺎس ‪ R6P‬أو ‪ LR6‬أو ‪(AA‬‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫اﻟذاﻛرة‬‫اﻟﺷﺎﺷﺔ‬‫رﻣز ﻋدم اﻧﺗظﺎم‬‫ﺿرﺑﺎت اﻟﻘﻠب‪/‬اﻟرﺟﻔﺎن اﻷذﯾﻧﻲ‬‫اﻟﺿﻐط‬‫اﻻﻧﻘﺑﺎﺿﻲ‬‫‪ %‬ﻋدم اﻧﺗظﺎم‬‫رﻣز ﺧطﺄ ﻓﻲ ﺗﺛﺑﯾت اﻟرﺑﺎط‬‫اﻟﺿﻐط‬‫اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﺗﺣرك‬‫ﻣؤﺷر ﺧطﺄ‬‫ّ‬‫ﻋﻼﻣﺔ اﻟﻘﻠب‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫وﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‬‫ﺳرﻋﺔ اﻟﻧﺑض‬‫ُ‬‫ﻣؤﺷر اﻟﺑطﺎرﯾﺔ‬‫اﻟﻌرﺑﯾﺔ ‪4‬‬

‫اﻟرﻣوز‬‫اﻟرﻣوز اﻟﺗﻲ ﯾﻣﻛن طﺑﺎﻋﺗﮭﺎ ﻋﻠﻰ ﻋﻠﺑﺔ اﻟﺟﮭﺎز‬‫اﻟرﻣوز‬‫‪SYS‬‬‫‪DIA‬‬‫‪PUL‬‬‫‪IP‬‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫وﺿﻊ اﻻﺳﺗﻌداد وﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫ﺿﻐط اﻟدم اﻻﻧﻘﺑﺎﺿﻲ ﺑﺎﻟﻣﻠﻠﯾﻣﺗر اﻟزﺋﺑﻘﻲ‬‫ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ ﺑﺎﻟﻣﻠﻠﯾﻣﺗر اﻟزﺋﺑﻘﻲ‬‫ﻋدد اﻟﻧﺑض ﻓﻲ اﻟدﻗﯾﻘﺔ‬‫دﻟﯾل ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت‬‫ﺗﯾﺎر ﻣﺳﺗﻣر‬‫اﻟﺗﻌرض‬‫ﺻﻣم اﻟﺟﮭﺎز واﻟرﺑﺎط واﻷﻧﺑوب ﻣن أﺟل ﺗوﻓﯾر ﺣﻣﺎﯾﺔ ﺧﺎﺻﺔ ﺿد‬‫ّ‬‫اﻟﻧوع ‪ُ :BF‬‬‫ﻟﻠﺻدﻣﺎت اﻟﻛﮭرﺑﺎﺋﯾﺔ‪.‬‬‫ﻣﻠﺻﻖ اﻟﺗوﺟﯾﮫ ‪ EC‬ﻟﻸﺟﮭزة اﻟطﺑﯾﺔ‬‫ﻣﻣﺛل اﻻﺗﺣﺎد اﻷوروﺑﻲ‬‫اﻟﺟﮭﺔ اﻟﻣﺻﻧّﻌﺔ‬‫ﺗﺎرﯾﺦ اﻟﺗﺻﻧﯾﻊ‬‫رﻣز اﻟﺣﻣﺎﯾﺔ اﻟدوﻟﯾﺔ‬‫رﻣز اﻟدﻟﯾل اﻟﺗوﺟﯾﮭﻲ ﻟﻧﻔﺎﯾﺎت اﻟﻣﻌدات اﻟﻛﮭرﺑﺎﺋﯾﺔ واﻹﻟﻛﺗروﻧﯾﺔ‬‫‪SN‬‬‫اﻟرﻗم اﻟﺗﺳﻠﺳﻠﻲ‬‫راﺟﻊ دﻟﯾل‪/‬ﻛﺗﯾّب اﻹرﺷﺎدات‬‫ﻗطﺑﯾﺔ ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫ﺣﺎﻓظ ﻋﻠﯾﮫ ﺟﺎﻓًﺎ‬‫اﻟرﻣوز اﻟﺗﻲ ﺗظﮭر ﻋﻠﻰ اﻟﺷﺎﺷﺔ‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﯾظﮭر ﺣﯾﻧﻣﺎ ﯾﻛون إﺟراء اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬ﯾوﻣض ﻋﻧد إﺟراء اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬ﯾظل ﺛﺎﺑﺗًﺎ‬‫ﻗدر اﻟﻣﺳﺗطﺎع‪.‬‬‫اﻟﻛﺷف ﻋن اﻟﻧﺑض‪.‬‬‫رﻣز ‪) IHB/AFib‬ﻋدم اﻧﺗظﺎم ﺿرﺑﺎت اﻟﻘﻠب‪/‬اﻟرﺟﻔﺎن‬‫اﻷذﯾﻧﻲ(‬‫ﯾظﮭر ﻋﻧد اﻟﻛﺷف ﻋن ﻋدم اﻧﺗظﺎم ﻓﻲ ﺿرﺑﺎت اﻟﻘﻠب‪.‬‬‫ﯾﻣﻛن أن ﯾُﺿﻲء ﻋﻧد اﻟﻛﺷف ﻋن وﺟود اھﺗزاز ﺑﺳﯾط‬‫ﻟﻠﻐﺎﯾﺔ‪ ،‬ﻣﺛل اﻻھﺗزاز أو اﻻرﺗﻌﺎش‪.‬‬‫ﯾﻣﻛن أن ﯾؤدي إﻟﻰ إظﮭﺎر ﻗﯾﻣﺔ‬‫ﺧﺎطﺋﺔ‪ .‬ﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس‬‫ﺗﺣرك اﻟﺟﺳم أو اﻟذراع أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ﯾظﮭر ﻋﻧد ّ‬‫ﻣرة أﺧرى‪.‬‬‫ُﻛن ﺛﺎﺑﺗًﺎ أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪5‬‬

‫اﻟرﻣوز‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﯾﻣﻛن أن ﯾؤدي إﻟﻰ إظﮭﺎر ﻗﯾﻣﺔ ﺧﺎطﺋﺔ‪.‬‬‫ﯾظﮭر أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس ﻓﻲ ﺣﺎل ﻋدم ﻟف اﻟرﺑﺎط‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪ ،‬وﺣﺎول‬‫ﺑﺈﺣﻛﺎم ﺣول اﻟذراع‬‫إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ﻣؤﺷر إﻟﻰ ﻣﻌدل ﻋدم اﻧﺗظﺎم ﺿرﺑﺎت‬‫اﻟﻘﻠب‪/‬اﻟرﺟﻔﺎن اﻷذﯾﻧﻲ اﻟ ُﻣﻛﺗﺷف ﻓﻲ اﻟذاﻛرة‬‫]‪x100 [%‬‬

‫ﻋدد ﺣﺎﻻت ‪ IHB/AFib‬اﻟ ُﻣﻛﺗﺷﻔﺔ‬‫ﻓﻲ اﻟذاﻛرة‬‫اﻟرﻗم اﻟﻛﻠﻲ‬‫= ‪%IHB / AFib‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس اﻟﺳﺎﺑﻘﺔ اﻟ ُﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‪.‬‬‫ﻣﺗوﺳط اﻟﺑﯾﺎﻧﺎت )ﻣﺗوﺳط ﻋدد اﻟﻘﯾﺎﺳﺎت اﻟﻣﺧزﻧﺔ(‬‫ﺑطﺎرﯾﺔ ﻛﺎﻣﻠﺔ اﻟﺷﺣن‬‫ﻣؤﺷر طﺎﻗﺔ اﻟﺑطﺎرﯾﺔ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‬‫ﺑطﺎرﯾﺔ ﻗﻠﯾﻠﺔ اﻟﺷﺣن‬‫ﺗﻛون طﺎﻗﺔ اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ ﻋﻧدﻣﺎ ﯾوﻣض اﻟرﻣز‪ .‬ﻋﻧدﻣﺎ ﯾوﻣض اﻟرﻣز‪.‬‬‫ﺿﻐط اﻟدم ﻏﯾر ﻣﺳﺗﻘر ﻧظرا ً‬‫ﻟﻠﺗﺣرك أﺛﻧﺎء إﺟراء ﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ّ‬‫اﻟﻘﯾﺎس‪.‬‬‫ُﻛن ﺛﺎﺑﺗًﺎ ﺗﻣﺎ ًﻣﺎ أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﻗﯾم اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ واﻻﻧﺑﺳﺎطﻲ ﺿﻣن ﻧطﺎق‬‫‪ 10‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ ﻣن ﻛﻠﯾﮭﻣﺎ‪.‬‬‫ﻗﯾﻣﺔ اﻟﺿﻐط ﻻ ﺗرﺗﻔﻊ أﺛﻧﺎء ﻧ ْﻔﺦ اﻟﮭواء‪.‬‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪ ،‬وﺣﺎول‬‫إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ﻟم ﯾُﻠف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﺧطﺄ ﻓﻲ ﻋرض ﺳرﻋﺔ اﻟﻧﺑض ﻓﻲ اﻟدﻗﯾﻘﺔ‬‫ﻟم ﯾﺗم اﻟﻛﺷف ﻋن اﻟﻧﺑض ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﺧطﺄ داﺧﻠﻲ ﻓﻲ ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫اﻟﻌرﺑﯾﺔ ‪6‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت واﺿﻐط ﻋﻠﻰ‬‫اﻟزر_‪ ، _START‬ﺛ ّم أ ِﻋد ﺗرﻛﯾب‬‫اﻟﺑطﺎرﯾﺎت ﻣرة أﺧرى‪ .‬اﺗﺻل ﺑﺎﻟوﻛﯾل‬‫إذا اﺳﺗﻣر ظﮭور اﻟﺧطﺄ‪.‬‬‫وﺿﻊ اﻟﺗﺷﻐﯾل‬‫‪ .1‬اﻟﻘﯾﺎس اﻟﻌﺎدي‬‫اﺿﻐط ﻋﻠﻰ اﻟزر _‪ . _START‬ﺗﺑدأ ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم وﺗُﺳ ّﺟل اﻟﺑﯾﺎﻧﺎت ﻓﻲ اﻟذاﻛرة‪ .‬ﺳﻌﺔ اﻟذاﻛرة ﻟﮭذا‬‫اﻟﺟﮭﺎز ﺗﺳﻣﺢ ﺑﺗﺧزﯾن آﺧر ‪ 60‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس‪.‬‬‫ﻓﻲ وﺿﻊ اﻻﺳﺗﻌداد‪،‬‬‫اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ‬‫‪ .2‬ﻛﯾﻔﯾﺔ ﻣراﺟﻌﺔ ﺑﯾﺎﻧﺎت اﻟذاﻛرة‬‫اﻟزر‬‫ﻋﻧد ﻋدم ظﮭور ﺷﻲء‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ اﻟزر _‪. _START‬‬‫ﺣرر اﻟزر ﻋﻧد ظﮭور ﻣﺗوﺳط اﻟﻘﯾﺎﺳﺎت اﻟﺳﺎﺑﻘﺔ اﻟﻣﺧزﻧﺔ‪.‬‬‫ّ‬‫ﺗُﻌرض ﻗراءات اﻟﻘﯾﺎﺳﺎت واﻟﻘﯾﺎﺳﺎت اﻟ ُﻣﺧزﻧﺔ ﺗﻠﻘﺎﺋﯾًﺎ ﺑﺎﻟﺗرﺗﯾب‬‫وذﻟك ﻣن آﺧر ﻣرة ﻗﯾﺎس‪.‬‬‫ﻟﻠﺣﺻول ﻋﻠﻰ ﺗﻔﺎﺻﯾل ﻛﯾﻔﯾﺔ اﻻطﻼع ﻋﻠﻰ اﻟﺑﯾﺎﻧﺎت‪ ،‬راﺟﻊ‬‫"اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة"‪.‬‬‫‪ .3‬ﺣذف ﺟﻣﯾﻊ اﻟﺑﯾﺎﻧﺎت اﻟﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‬‫ﻋﻨﺪ إﻏﻼق ا�جهﺎز‪ ،‬اﺿﻐﻂ ﻋ�� اﻟﺰر‬‫_‪� _START‬ﺸ�ﻞ ﻣﺴﺘﻤﺮ ﺣ�ى ﻳﻈهﺮ اﻟﺮﻣﺰ "‬‫ﻋ�� اﻟﺸﺎﺷﺔ‪.‬‬‫" ﳌ�ح اﻟﺒﻴﺎﻧﺎت‪.‬‬‫ﺣﺪد "‬‫‪.‬‬‫ﻳﺘﻢ ﻣ�ح اﻟﺒﻴﺎﻧﺎت ﻋﻨﺪﻣﺎ ﻳﻮﻣﺾ اﻟﺮﻣﺰ‬‫ً‬‫ﺗﻠﻘﺎﺋﻴﺎ‪.‬‬‫ﻳﻨﻄﻔﺊ ا�جهﺎز‬‫"‬‫‪ .4‬إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ‬‫‪START‬‬‫راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪ 14‬ﻟﻠﺗﻌرف ﻋﻠﻰ إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣطﻠوب‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪7‬‬‫‪START‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﺗرﻛﯾب ‪ /‬ﺗﻐﯾﯾر اﻟﺑطﺎرﯾﺎت‬‫اﻟﺧطوة ‪1‬‬‫‪ .1‬اﻧزع ﻏطﺎء اﻟﺑطﺎرﯾﺎت‪.‬‬‫‪ .2‬اﻧزع اﻟﺑطﺎرﯾﺎت اﻟﻣﺳﺗﺧدﻣﺔ وأدﺧل اﻷﺧرى اﻟﺟدﯾدة‬‫ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت ﻛﻣﺎ ھو ﻣوﺿّﺢ‪ ،‬ﻣﻊ ﻣراﻋﺎة‬‫اﻻﺗﺟﺎه اﻟﺻﺣﯾﺢ ﻟﻌﻼﻣﺗﻲ اﻟﻘطﺑﯾﺔ )‪ +‬و ‪.(-‬‬‫ﻻ ﺗﺳﺗﺧدم ﺳوى ﺑطﺎرﯾﺎت ﻣﻘﺎس ‪ R6P‬أو ‪ LR6‬أو ‪.AA‬‬‫‪ .3‬ر ّﻛب ﻏطﺎء اﻟﺑطﺎرﯾﺎت‪.‬‬‫اﻟﺧطوة ‪3‬‬‫اﻟﺧطوة ‪1‬‬‫اﻟﺧطوة ‪2‬‬‫ﺗﻧﺑﯾﮫ‬‫أدﺧل اﻟﺑطﺎرﯾﺎت اﻟﺟدﯾدة ﻛﻣﺎ ھو ﻣوﺿّﺢ ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت‪ .‬وﻟن ﯾﻌﻣل اﻟﺟﮭﺎز ﻓﻲ ﺣﺎﻟﺔ ﺗرﻛﯾﺑﮭﺎ ﺑطرﯾﻘﺔ‬‫ﻏﯾر ﺻﺣﯾﺣﺔ‪.‬‬‫)رﻣز اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ( ﻋﻠﻰ اﻟﺷﺎﺷﺔ‪ ،‬اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪ .‬ﻻ‬‫ﻋﻧدﻣﺎ ﯾوﻣض‬‫ﺗﻌرض‬‫ﺗﺳﺗﺧدم ﺑطﺎرﯾﺎت ﺟدﯾدة ﻣﻊ أﺧرى ﻗدﯾﻣﺔ‪ .‬ﻗد ﯾﺗﺳﺑب ذﻟك ﻓﻲ ﺗﻘﺻﯾر ﻓﺗرة اﺳﺗﺧدام اﻟﺑطﺎرﯾﺎت‪ ،‬أو ﻓﻲ ّ‬‫اﻟﺟﮭﺎز ﻟﻠﻌطل‪.‬‬‫اﺳﺗﺑدل اﻟﺑطﺎرﯾﺎت ﺑﻌد إﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز ﺑﺛﺎﻧﯾﺗﯾن أو أﻛﺛر‪.‬‬‫ﻓﺄﺟر ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫)رﻣز اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ( ﺣﺗﻰ ﺑﻌد اﺳﺗﺑدال اﻟﺑطﺎرﯾﺎت‪،‬‬‫إذا ظﮭر‬‫ِ‬‫ﯾﺗﻌرف اﻟﺟﮭﺎز ﺑﻌدﺋ ٍذ ﻋﻠﻰ اﻟﺑطﺎرﯾﺎت اﻟﺟدﯾدة‪.‬‬‫ﯾﻣﻛن أن‬‫ّ‬‫ﻻ ﯾظﮭر )رﻣز اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ( ﻋﻧد ﻧﻔﺎد طﺎﻗﺔ اﻟﺑطﺎرﯾﺎت‪.‬‬‫ﺗﺗﺑﺎﯾن ﻓﺗرة ﺗﺷﻐﯾل اﻟﺑطﺎرﯾﺎت ﺣﺳب درﺟﺔ اﻟﺣرارة اﻟﻣﺣﯾطﺔ‪ ،‬وﻗد ﺗﻛون ﻓﺗرة ﺗﺷﻐﯾﻠﮭﺎ أﻗﺻر ﻋﻧد درﺟﺎت‬‫اﻟﺣرارة اﻟﻣﻧﺧﻔﺿﺔ‪ .‬وﺑﺷﻛل ﻋﺎم‪ ،‬ﺳﺗدوم اﻟﺑطﺎرﯾﺎت اﻷرﺑﻊ اﻟﺟدﯾدة ﻣﻘﺎس ‪ LR6‬ﻣدة ﻋﺎم واﺣد ﺗﻘرﯾﺑًﺎ ﻋﻧد‬‫اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻠﻘﯾﺎس ﻣرﺗﯾن ﻛل ﯾوم‪.‬‬‫ﻻ ﺗﺳﺗﺧدم ﺳوى اﻟﺑطﺎرﯾﺎت اﻟﻣﺣددة ﻓﻘط‪ .‬وﺗﻛون اﻟﺑطﺎرﯾﺎت اﻟﻣرﻓﻘﺔ ﻣﻊ اﻟﺟﮭﺎز ﻷﻏراض اﺧﺗﺑﺎر أداء ﺟﮭﺎز‬‫اﻟﻘﯾﺎس‪ ،‬وﯾﻣﻛن أن ﺗﻛون ذات ﻓﺗرة ﺗﺷﻐﯾل ﻣﺣدودة‪.‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت ﻓﻲ ﺣﺎﻟﺔ ﻋدم اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻔﺗرة طوﯾﻠﺔ‪.‬‬‫ﺗﺗﻌرض اﻟﺑطﺎرﯾﺎت ﻟﻠﺗﺳرﯾب ﻣﻣﺎ ﯾﺗﺳﺑب ﻓﻲ ّ‬‫ﺗﻌطل اﻟﺟﮭﺎز‪.‬‬‫ﯾﻣﻛن أن‬‫ّ‬‫اﻟﻌرﺑﯾﺔ ‪8‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﻣﻘﺑس اﻟﮭواء‬‫ﺗوﺻﯾل ﺧرطوم اﻟﮭواء‬‫ﺻل اﻟﮭواء ﻓﻲ ﻣﻘﺑس اﻟﮭواء‬‫أدﺧل ﻗﺎﺑس ﻣو ّ‬‫ﺑﺈﺣﻛﺎم‪.‬‬‫ﺻل اﻟﮭواء‬‫ﻗﺎﺑس ﻣو ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﺗوﺻﯾل‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻓﻲ ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‪.‬‬‫أدﺧل ﻗﺎﺑس ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﺑﻣﺄﺧذ اﻟﺗﯾﺎر اﻟﻛﮭرﺑﺎﺋﻲ‪.‬‬‫ﺑﻌد ذﻟك‪ِ ،‬‬‫ﺻل ّ‬‫ﻣﻘﺑس اﻟﺗﯾﺎر‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻗﺎﺑس ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد اﻟﻣﺣدد‪.‬‬‫اﺳﺗﺧدم ّ‬‫)راﺟﻊ اﻟﺻﻔﺣﺔ ‪(.21‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد واﺳﺣﺑﮫ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻣن ﻣﺄﺧذ اﻟﺗﯾﺎر اﻟﻛﮭرﺑﺎﺋﻲ‪ ،‬أﻣﺳك ھﯾﻛل‬‫ﻋﻧد ﻓﺻل‬‫ّ‬‫ّ‬‫ﺧﺎرج اﻟﻣﺄﺧذ‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻣن ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬أﻣﺳك‬‫ﻋﻧد ﻓﺻل‬‫ّ‬‫ّ‬‫واﺳﺣﺑﮫ ﺧﺎرج اﻟﺟﮭﺎز‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪9‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﺧﺗﯾﺎر ﻣﻘﺎس رﺑﺎط اﻟذراع اﻟﺻﺣﯾﺢ‬‫إن اﺳﺗﻌﻣﺎل اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟرﺑﺎط اﻟذراع ھﺎم ﺟداً ﻓﻲ د ّﻗﺔ ﻗﯾﺎس ﺿﻐط اﻟدم‪ .‬وإذا ﻟم ﯾﻛن ﻣﻘﺎس اﻟرﺑﺎط‬‫ﺻﺣﯾ ًﺣﺎ‪ ،‬ﻓﻘﯾﺎس ﺿﻐط اﻟدم ﺳﯾظﮭر ﻗﯾﻣﺔ ﻏﯾر ﺻﺣﯾﺣﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﻋﺎ ﻋﻠﻰ ﻛل رﺑﺎط‪.‬‬‫ﯾﻛون ﻣﻘﺎس ﻣﺣﯾط اﻟذراع ﻣطﺑو ً‬‫ﯾﺳﺎﻋد ﻛ ﱞل ﻣن اﻟﻣؤﺷر واﻟﻧطﺎق اﻟﻣﻧﺎﺳب‪ ،‬اﻟﻣطﺑوﻋﺎن ﻋﻠﻰ اﻟرﺑﺎط‪ ،‬ﻋﻠﻰ أﻧك ﺗﺳﺗﺧدم اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ‪.‬‬‫)راﺟﻊ "اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ اﻟرﺑﺎط" ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ(‬‫إذا ﻛﺎن اﻟﻣؤﺷر ﯾﺷﯾر إﻟﻰ ﺧﺎرج اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل اﻟﻣﺣﻠﻲ ﻟﺷراء رﺑﺎط ﺑدﯾل‪.‬‬‫رﺑﺎط اﻟذراع ھو أﺣد اﻟﻣواد اﻟﻣﺳﺗﮭﻠﻛﺔ‪ .‬ﻓﻲ ﺣﺎل أﺻﺑﺢ اﻟرﺑﺎط ﺑﺎﻟﯾﺎ ً ﻓﺎﺳْﺗﺑدﻟﮫ ﻓورا ً ﺑﺂﺧر ﺟدﯾد‪.‬‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫ﺣﺟم اﻟرﺑﺎط اﻟﻣوﺻﻰ ﺑﮫ‬‫ﺣﺟم اﻟذراع‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬‫‪ 31‬ﺳم إﻟﻰ ‪ 45‬ﺳم‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬‫‪ 22‬ﺳم إﻟﻰ ‪ 42‬ﺳم‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪ 22‬ﺳم إﻟﻰ ‪ 32‬ﺳم‬‫ﺣﺟم اﻟذراع‪ :‬اﻟﻣﺣﯾط ﻋﻧد ﻋﺿﻠﺔ اﻟذراع‪.‬‬‫ﻟم ﯾُﺻﻣم اﻟﺟﮭﺎز ‪ UA-767S-W‬ﻟﻐرض اﺳﺗﺧدام ﻣﻘﺎس رﺑﺎط ذراع ﺻﻐﯾر‪.‬‬‫ﻣﻼﺣظﺔ‪:‬‬‫ﻛﯾﻔﯾﺔ وﺿﻊ رﺑﺎط اﻟذراع‬‫ﺿد )أﻋﻠﻰ اﻟذراع(‪ ،‬ﻓوق اﻟﺟزء‬‫‪ .1‬ﻟف اﻟرﺑﺎط ﺣول اﻟﻌ ُ‬‫اﻟداﺧﻠﻲ ﻣن اﻟﻣرﻓﻖ ﺑﻣﻘدار‪ 1-2‬ﺳم ﺗﻘرﯾﺑًﺎ‪ ،‬ﻛﻣﺎ ھو‬‫ﻣوﺿّﺢ ﻓﻲ اﻟﺷﻛل‪.‬‬‫ﯾﺟب وﺿﻊ اﻟرﺑﺎط ﻋﻠﻰ اﻟﺟﻠد ﻣﺑﺎﺷرة‪ ،‬وﻟﯾس ﻋﻠﻰ‬‫اﻟﻣﻼﺑس ﻣﺑﺎﺷرة ﻷﻧﮭﺎ ﺗﺗﺳﺑب ﺑﻘﯾﺎس ﻧﺑﺿﺎت ﺿﻌﯾﻔﺔ‪،‬‬‫ﻣﻣﺎ ﯾؤدي إﻟﻰ ﺧطﺄ ﻓﻲ اﻟﻘﯾﺎس‪.‬‬‫ﻲ‬‫‪ .2‬ﯾﻣﻛن أن ﯾﻣﻧﻊ اﻟﺗﺿﯾﯾﻖ اﻟﺷدﯾد ﻋﻠﻰ اﻟﻌ ُ‬‫ﺿد‪ ،‬ﺑﺳﺑب ط ّ‬‫ﻛم اﻟﻘﻣﯾص ﺑﺷدة‪ ،‬اﻟﺣﺻول ﻋﻠﻰ ﻗراءات دﻗﯾﻘﺔ‪.‬‬‫‪ .3‬ﺗﺄﻛد ﻣن وﺟود ﻧﻘﺎط اﻟﻣؤﺷر‬‫اﻟﻣﻧﺎﺳب‪.‬‬‫ﺿﻣن اﻟﻧطﺎق‬‫ﻲ ﻛم اﻟﻘﻣﯾص ﺑﺷدة‪.‬‬‫ﻻ ﺗﻘم ﺑط ّ‬‫اﻟرﺑﺎط‬‫‪ 1-2‬ﺳم‬‫ﺧرطوم‬‫ﻋﻼﻣﺔ ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫أﺣﻛم ﺗﺛﺑﯾت اﻟﻣﺛﺑت‬‫اﻟﻘﻣﺎﺷﻲ‪.‬‬‫اﻟﻣؤﺷر‬‫اﻟﻌرﺑﯾﺔ ‪10‬‬‫واﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ اﻟرﺑﺎط‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﺿﺑط اﻟﻌﻼﻣﺔ ﺑﺣﯾث ﺗﻛون ﻋﻧد‬‫ﺿد أو ﻓﻲ ﻣﺣﺎذاة ﻣﻊ إﺻﺑﻊ‬‫ﺷرﯾﺎن اﻟﻌ ُ‬‫اﻟﺑﻧﺻر ﻋﻧد اﻟﺟزء اﻟداﺧﻠﻲ ﻣن‬‫اﻟذراع‪.‬‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﻋﻼﻣﺔ ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫▲‬‫اﻟﻣؤﺷر‬‫‪REF‬‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟرﺑﺎط اﻟﺑﺎﻟﻐﯾن‪.‬‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن‪.‬‬‫ﯾﻛون ﻣطﺑو ً‬‫ﺗﺟﺎوز اﻟﻧطﺎق اﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن واﻟرﺑﺎط‬‫ُ‬‫ذي اﻟﻧطﺎق اﻟﻌرﯾض‪.‬‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟﻠرﺑﺎط ذي اﻟﻧطﻖ اﻟﻌرﯾض‪ .‬ﯾﻛون‬‫ﻣطﺑو ًﻋﺎ ﻋﻠﻰ اﻟرﺑﺎط ذي اﻟﻧطﻖ اﻟﻌرﯾض‪.‬‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟرﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪ .‬ﯾﻛون‬‫ﻣطﺑو ًﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫أدﻧﻰ اﻟﻧطﺎق اﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن واﻟرﺑﺎط ذي‬‫اﻟﻧطﺎق اﻟﻌرﯾض‪.‬‬‫أدﻧﻰ اﻟﻧطﺎﻗﺎﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر ﺑد ً‬‫ﻻ ﻣن‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن أو اﻟرﺑﺎط ذي اﻟﻧطﺎق‬‫اﻟﻌرﯾض‪.‬‬‫اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن ﺑدﻻً ﻣن رﺑﺎط‬‫اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫رﻗم اﻟﺗﺷﻐﯾﻠﺔ‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬

‫‪L‬‬

‫‪A‬‬

‫‪L‬‬

‫‪W‬‬

‫‪L‬‬

‫‪A‬‬

‫اﻟﻌرﺑﯾﺔ ‪11‬‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬‫‪S‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪S‬‬

‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﻛﯾﻔﯾﺔ إﺟراء ﻋﻣﻠﯾﺎت ﻗﯾﺎس دﻗﯾﻘﺔ‬‫ﻟﻠﺣﺻول ﻋﻠﻰ ﻗﯾﺎس دﻗﯾﻖ ﺗﻣﺎ ًﻣﺎ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪:‬‬‫ﻖ ﻗدﻣﯾك ﻣﺳﺗوﯾﺗﯾن‬‫اﺟﻠس ﻣﺳﺗرﯾ ًﺣﺎ ﻋﻠﻰ ﻛرﺳﻲ‪.‬‬‫أرخ ذراﻋك ﻋﻠﻰ اﻟطﺎوﻟﺔ‪ .‬ﻻ ﺗﺿﻊ ﻗد ًﻣﺎ ﻓوق اﻷﺧرى‪ .‬أﺑ ِ‬‫ِ‬‫ﻋﻠﻰ اﻷرض واﺟﻌل ظﮭرك ﻣﺳﺗﻘﯾ ًﻣﺎ‪.‬‬‫اﺳﺗرخ ﻟﻣدة ﺗﺗراوح ﺑﯾن ﺧﻣس إﻟﻰ ﻋﺷر دﻗﺎﺋﻖ ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫ِ‬‫اﺟﻌل ﻣﻧﺗﺻف اﻟرﺑﺎط ﻋﻧد اﻟﻣﺳﺗوى ﻧﻔﺳﮫ ﻟﻠﻘﻠب‪.‬‬‫اﺑﻖ ﺛﺎﺑﺗًﺎ وﺣﺎﻓظ ﻋﻠﻰ ھدوءك أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ﻻ ﺗﻘس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓور اﻻﻧﺗﮭﺎء ﻣن ﻣﻣﺎرﺳﺔ اﻟﺗﻣرﯾﻧﺎت اﻟﺑدﻧﯾﺔ أو اﻻﺳﺗﺣﻣﺎم‪ .‬اﺳﺗرخ ﻟﻣدة ﺗﺗراوح‬‫ﺑﯾن ﻋﺷرﯾن وﺛﻼﺛﯾن دﻗﯾﻘﺔ ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫ﺣﺎول ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ ﻛل ﯾوم‪.‬‬‫اﻟﻘﯾﺎس‬‫أﺛﻧﺎء اﻟﻘﯾﺎس‪ ،‬ﻣن اﻟطﺑﯾﻌﻲ أن ﺗﺷﻌر أن اﻟرﺑﺎط ﻣﺷدود ﻟﻠﻐﺎﯾﺔ‪) .‬ﻻ ﺗﻠﺗﻔت ﻟذﻟك‪(.‬‬‫ﺑﻌد اﻟﻘﯾﺎس‬‫ﺑﻌد اﻟﻘﯾﺎس‪ ،‬اﺿﻐط ﻋﻠﻰ اﻟزر _‪ _START‬ﻹﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫اﻧزع اﻟرﺑﺎط وﺳﺟل ﺑﯾﺎﻧﺎﺗك‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬ﯾﺗﻣﯾّز اﻟﺟﮭﺎز ﺑوظﯾﻔﺔ إﯾﻘﺎف اﻟﺗﺷﻐﯾل اﻟﺗﻠﻘﺎﺋﻲ‪ ،‬ﺣﯾث ﯾﻧطﻔﺊ اﻟﺟﮭﺎز ﺑﻌد ُﻣﺿ ّﻲ دﻗﯾﻘﺔ واﺣد ﺗﻘرﯾﺑًﺎ ﻣن اﻻﻧﺗﮭﺎء ﻣن‬‫ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫اﺟﻌل اﻟﻣدة اﻟزﻣﻧﯾﺔ اﻟﻔﺎرﻗﺔ ﺑﯾن ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻧﻔس اﻟﺷﺧص ﺛﻼث دﻗﺎﺋﻖ ﻋﻠﻰ اﻷﻗل‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪12‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‬‫ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ ،‬راﺟﻊ "ﻣﻼﺣظﺎت ﺑﺎﻟﻧﺳﺑﺔ ﻟﻠﻘﯾﺎس اﻟدﻗﯾﻖ" ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫اﻟﻘﯾﺎس اﻟﻌﺎدي‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫‪.1‬‬‫ﻟف اﻟرﺑﺎط ﺣول اﻟذراع )ﯾُﻔﺿل اﻟذراع اﻷﯾﺳر(‪.‬‬‫اﺟﻠس ھﺎدﺋًﺎ أﺛﻧﺎء إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر ‪. _START‬‬‫ﯾﺗم ﻋرض ﺟﻣﯾﻊ ﺷراﺋﺢ اﻟﺷﺎﺷﺔ‪.‬‬‫ﯾوﻣض اﻟﺻﻔر )‪ (0‬ﻋﻠﻰ اﻟﺷﺎﺷﺔ ﻟﻔﺗرة وﺟﯾزة‪.‬‬‫ﺗﺗﻐﯾّر اﻟﺷﺎﺷﺔ‪ ،‬ﻛﻣﺎ ھو ﻣوﺿّﺢ ﻓﻲ اﻟﺷﻛل ﻋﻠﻰ اﻟﯾﻣﯾن‪،‬‬‫ﺣﯾث ﺗﺑدأ ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬ﯾﺑدأ اﻟرﺑﺎط ﻓﻲ اﻻﻧﺗﻔﺎخ‪ .‬ﻣن‬‫اﻟطﺑﯾﻌﻲ أن ﺗﺷﻌر أن اﻟرﺑﺎط ﻣﺷدود ﻟﻠﻐﺎﯾﺔ‪ .‬ﯾُﻌرض‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‪ ،‬ﻋﻠﻰ اﻟﺣﺎﻓﺔ اﻟﯾﺳرى ﻣن اﻟﺷﺎﺷﺔ‪،‬‬‫أﺛﻧﺎء ﻧ ْﻔﺦ اﻟﮭواء‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬إذا رﻏﺑت ﻓﻲ إﯾﻘﺎف ﻧ ْﻔﺦ اﻟﮭواء ﻓﻲ أي وﻗت‪،‬‬‫ﻓﺎﺿﻐط ﻋﻠﻰ اﻟزر _‪ START‬ﻣرة أﺧرى‪.‬‬‫‪.3‬‬‫ﻋﻧد اﻛﺗﻣﺎل ﻧ ْﻔﺦ اﻟﮭواء‪ ،‬ﺗﺑدأ ﻋﻣﻠﯾﺔ ﺗﻔرﯾﻎ اﻟﮭواء ﺗﻠﻘﺎﺋﯾًﺎ‬‫)رﻣز اﻟﻘﻠب(‪ ،‬ﻣﺎ ﯾﺷﯾر إﻟﻰ أن ﻋﻣﻠﯾﺔ‬‫وﯾوﻣض‬‫اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬وﯾوﻣض رﻣز اﻟﻘﻠب‪ ،‬ﻓور اﻛﺗﺷﺎف‬‫وﺟود اﻟﻧﺑض‪ ،‬ﻣﻊ ﻛل ﻧﺑﺿﺔ‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬ﻓﻲ ﺣﺎﻟﺔ ﻋدم اﻟﺣﺻول ﻋﻠﻰ ﺿﻐط ﻣﻼﺋم‪ ،‬ﯾﺑدأ‬‫اﻟﺟﮭﺎز ﻓﻲ ﻧ ْﻔﺦ اﻟﮭواء ﻣرة أﺧرى ﺗﻠﻘﺎﺋﯾًﺎ‪.‬‬‫وﻟﺗﻔﺎدي إﻋﺎدة اﻟﻧﻔﺦ‪ ،‬راﺟﻊ "إﺟراء اﻟﻘﯾﺎس ﻣﻊ‬‫اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ" ﻓﻲ‬‫اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫‪.4‬‬‫ﻋﻧد اﻛﺗﻣﺎل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ ،‬ﺗُﻌرض ﻗراءات‬‫ﺳرﻋﺔ اﻟﻧﺑض‪.‬‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ واﻻﻧﺑﺳﺎطﻲ و ُ‬‫ﯾﻔرغ اﻟرﺑﺎط اﻟﮭواء اﻟﻣﺗﺑﻘﻲ ﺗﻣﺎ ًﻣﺎ‪.‬‬‫ّ‬‫إﺟراء اﻟﻘﯾﺎس ﻗﯾد‬‫اﻟﺗﻧﻔﯾذ‬‫‪.5‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر _‪ _START‬ﻹﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫ﯾﻧطﻔﺊ اﻟﺟﮭﺎز ﺗﻠﻘﺎﺋﯾًﺎ ﺑﻌد ُﻣﺿﻲ دﻗﯾﻘﺔ واﺣدة ﻣن ﺣﺎﻟﺔ ﻋدم‬‫اﻟﺗﺷﻐﯾل‪.‬‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫ﻣﻼﺣظﺔ‪ :‬اﺟﻌل اﻟﻣدة اﻟزﻣﻧﯾﺔ اﻟﻔﺎرﻗﺔ ﺑﯾن ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻧﻔس‬‫اﻟﺷﺧص ﺛﻼث دﻗﺎﺋﻖ ﻋﻠﻰ اﻷﻗل‪.‬‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ‬‫اﻟﻌﺎﻟﻣﯾﺔ‬‫‪START‬‬‫اﺿﻐط‬‫ﻋرض ﺟﻣﯾﻊ ﺷراﺋﺢ‬‫اﻟﺷﺎﺷﺔ‬‫ﻋرض ﺻﻔر‬‫ﺑدء ﻧ ْﻔﺦ اﻟﮭواء‬‫ﺟﺎر ﻗﯾﺎس اﻟﺿﻐط‬‫ٍ‬‫ﺗﻔرﯾﻎ اﻟﮭواء اﻟﻣﺗﺑﻘﻲ‬‫ﺗﻠﻘﺎﺋﯾًﺎ‬‫اﻟﻌرﺑﯾﺔ ‪13‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‬‫إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ‬‫ﺻﻣم اﻟﺟﮭﺎز ‪ UA-767S-W‬ﻻﻛﺗﺷﺎف اﻟﻧﺑض وﻧ ْﻔﺦ اﻟرﺑﺎط ﺑﺎﻟﮭواء ﻋﻧد ﻣﺳﺗوى اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫ُ‬‫ﺗﻠﻘﺎﺋﯾًﺎ‪.‬‬‫اﺳﺗﺧدم ھذه اﻟطرﯾﻘﺔ ﻋﻧد ﺗﻛرار ﻧ ْﻔﺦ اﻟﮭواء أو ﻋﻧد ﻋدم ﻋرض اﻟﻧﺗﺎﺋﺞ ﺣﺗﻰ ﻋﻧد اﻧﺧﻔﺎض ﻣﺳﺗوى اﻟﺿﻐط‬‫إﻟﻰ ‪ 20‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ أو أﻗل‪.‬‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫‪.1‬‬‫ﻟف اﻟرﺑﺎط ﺣول اﻟذراع ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب )ﯾُﻔﺿل اﻟذراع‬‫اﻷﯾﺳر(‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر _‪. _START‬‬‫‪.3‬‬‫ﻋﻧدﻣﺎ ﯾوﻣض اﻟﺻﻔر‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ اﻟزر‬‫_‪ START‬إﻟﻰ أن ﯾظﮭر رﻗم ﯾﺗراوح ﺑﯾن ‪ 30‬و‪40‬‬‫ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ أﻋﻠﻰ ﻣن اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟذي‬‫ﺗﺗوﻗﻌﮫ‪.‬‬‫ﺣرر اﻟزر ﻋﻧد ﻣﺳﺗوى‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣطﻠوب‬‫‪.4‬‬‫ﻋﻧد ﺑﻠوغ اﻟرﻗم اﻟﻣطﻠوب‪ ،‬ﺣرر اﻟزر _‪ START‬ﻟﺑدء‬‫اﻟﻘﯾﺎس‪ .‬ﺗﺎﺑﻊ إﺟراء ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻛﻣﺎ ھو‬‫ﻣوﺿﺢ ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺳﺎﺑﻘﺔ‪.‬‬‫اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ‬‫اﻟزر‬‫‪ START‬اﺿﻐط ﻋﻠﻰ‬‫اﻟزر‬‫‪START‬‬‫راﺟﻊ اﻟﺻﻔﺣﺔ اﻟﺳﺎﺑﻘﺔ ﻟﻼطﻼع ﻋﻠﻰ‬‫إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﻣﻼﺣظﺎت ﺑﺎﻟﻧﺳﺑﺔ ﻟﻠﻘﯾﺎس اﻟدﻗﯾﻖ‬‫اﺟﻠس ﻓﻲ وﺿﻊ ﻣرﯾﺢ‪ .‬ﺿﻊ ذراﻋك ﻋﻠﻰ طﺎوﻟﺔ ﺑﺣﯾث ﺗﻛون راﺣﺔ ﯾدك ﻣواﺟﮭﺔ ﻷﻋﻠﻰ واﻟرﺑﺎط ﻋﻧد‬‫ﻣﺳﺗوى اﻟﻘﻠب‪.‬‬‫اﺳﺗرخ ﻟﻣدة ﺗﺗراوح ﺑﯾن ﺧﻣس إﻟﻰ ﻋﺷر دﻗﺎﺋﻖ ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬إذا ﻛﺎﻧت ﺗﻌﺗرﯾك ﻓرﺣﺔ ﻋﺎرﻣﺔ أو‬‫ِ‬‫ﺗﻣر ﺑﺣﺎﻟﺔ اﻛﺗﺋﺎب ﻧﺗﯾﺟﺔ ﻟﻠﺿﻐوطﺎت اﻟﻌﺎطﻔﯾﺔ‪ ،‬ﻓﺳﯾﻧﻌﻛس ذﻟك ﻣﻧﺗ ًﺟﺎ ﻗراءة أﻋﻠﻰ )أو أﻗل( ﻣن اﻟﻘراءة‬‫ﺳرﻋﺔ اﻟﻧﺑض أﺳرع ﻣن اﻟﻣﻌدل اﻟطﺑﯾﻌﻲ‪.‬‬‫اﻟطﺑﯾﻌﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬وﺳﺗﻛون ﻗراءة ُ‬‫ﯾﺗﺑﺎﯾن ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟﻠﻔرد ﺑﺎﺳﺗﻣرار ﺗﺑﻌًﺎ ﻟﻠﻌﻣل اﻟذي ﺗﻘوم ﺑﮫ واﻟطﻌﺎم اﻟذي ﺗﺗﻧﺎوﻟﮫ‪ .‬وﯾﻛون‬‫ﻛﺑﯾرا ﻟﻠﻐﺎﯾﺔ وﺳرﯾﻌًﺎ ﻋﻠﻰ ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﺗﺄﺛﯾرا‬‫ﺿﺎ‬‫ﻟﻠﻣﺷروﺑﺎت اﻟﺗﻲ ﺗﺗﻧﺎوﻟﮭﺎ أﯾ ً‬‫ً‬‫ً‬‫ﯾﻌﺗﻣد ھذا اﻟﺟﮭﺎز ﻓﻲ ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس اﻟﺗﻲ ﯾُﺟرﯾﮭﺎ ﻋﻠﻰ ﺿرﺑﺎت اﻟﻘﻠب‪ .‬ﻓﺈذا ﻛﺎن ﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟدﯾك‬‫ﺿﻌﯾﻔًﺎ ﺟدًا أو ﻏﯾر ﻣﻧﺗظم‪ ،‬ﻓﻘد ﯾﺟد اﻟﺟﮭﺎز ﺻﻌوﺑﺔ ﻓﻲ ﺗﺣدﯾد ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪.‬‬‫إذا اﻛﺗﺷف اﻟﺟﮭﺎز ﺣﺎﻟﺔ ﻏﯾر طﺑﯾﻌﯾﺔ‪ ،‬ﻓﺳﯾوﻗف ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس وﯾُظﮭر رﻣز اﻟﺧطﺄ‪ .‬راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪ 6‬ﻟﻼطﻼع ﻋﻠﻰ‬‫وﺻف اﻟرﻣوز‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪14‬‬‫ھذا اﻟﺟﮭﺎز ُﻣﺧﺻص ﻟﻼﺳﺗﺧدام ﻣن ﻗِﺑل اﻟﺑﺎﻟﻐﯾن‪ .‬اﺳﺗﺷر طﺑﯾﺑك ﻗﺑل اﺳﺗﺧدام ھذا اﻟﺟﮭﺎز ﻣﻊ أﺣد اﻷطﻔﺎل‪.‬‬‫ﯾﺟب ﻋدم اﺳﺗﺧدام اﻷطﻔﺎل اﻟﺟﮭﺎز دون إﺷراف‪.‬‬‫ﯾﻣﻛن أن ﯾﺗﺄﺛر أداء ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﺗﻠﻘﺎﺋﻲ ﺑﺳﺑب اﻟزﯾﺎدة اﻟﺑﺎﻟﻐﺔ ﻓﻲ درﺟﺔ اﻟﺣرارة أو‬‫اﻟرطوﺑﺔ أو اﻻرﺗﻔﺎع ﻋن ﺳطﺢ اﻟﺑﺣر‪.‬‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة‬‫ﻣﻼﺣظﺔ‪ :‬ﯾﺧزن ھذا اﻟﺟﮭﺎز آﺧر ‪ 60‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻓﻲ اﻟذاﻛرة‪.‬‬‫‪.1‬‬‫ﻋﻧد ﻋدم ﻋرض ﺷﻲء‪ ،‬اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ‬‫اﻟزر _‪ _ START‬ﻻﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت اﻟﻣﺧزﻧﺔ‪.‬‬‫‪.2‬‬‫ﺣرر اﻟزر ﻋﻧد ظﮭور ﻣﺗوﺳط اﻟﻘﯾﺎﺳﺎت اﻟﺳﺎﺑﻘﺔ‬‫ّ‬‫اﻟﻣﺧزﻧﺔ‪.‬‬‫‪.3‬‬‫ﺗُﻌرض ﻗراءات اﻟﻘﯾﺎﺳﺎت واﻟﻘﯾﺎﺳﺎت اﻟ ُﻣﺧزﻧﺔ ﺗﻠﻘﺎﺋﯾًﺎ‬‫ﺑﺎﻟﺗرﺗﯾب وذﻟك ﻣن آﺧر ﻣرة ﻗﯾﺎس‪.‬‬‫‪.4‬‬‫ﺳﺗﻧطﻔﺊ اﻟﺷﺎﺷﺔ ﺗﻠﻘﺎﺋﯾًﺎ ﺑﻌد ﻋرض ﺟﻣﯾﻊ اﻟﺑﯾﺎﻧﺎت‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬إذا ﺿﻐطت ﻋﻠﻰ اﻟزر _‪ _ START‬أﺛﻧﺎء اﺳﺗدﻋﺎء‬‫اﻟﺑﯾﺎﻧﺎت‪ ،‬ﻓﺳﯾﻧطﻔﺊ اﻟﺟﮭﺎز‪.‬‬‫اﻟزر‬‫‪ START‬اﺿﻐط ﺑﺎﺳﺗﻣرار‬‫ﻋﻠﻰ اﻟزر‬‫اﻟﻣﺗوﺳط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﻣﺗوﺳط اﻻﻧﺑﺳﺎطﻲ‬‫ﻣﺗوﺳط اﻟﻧﺑﺿﺎت‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫ﻟﺟﻣﯾﻊ اﻟﺑﯾﺎﻧﺎت‬‫اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﻧﺑﺿﺎت‬‫أﺣدث اﻟﺑﯾﺎﻧﺎت‬‫اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﻧﺑﺿﺎت‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻘدﯾﻣﺔ‬‫)اﻷﺧﯾرة(‬‫اﻟﻌرﺑﯾﺔ ‪15‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣؤﺷر ‪) IHB/AFib‬ﻋدم اﻧﺗظﺎم‬‫ﺿرﺑﺎت اﻟﻘﻠب‪/‬اﻟرﺟﻔﺎن اﻷذﯾﻧﻲ(؟‬‫ﻋﻧد اﻛﺗﺷﺎف ﺟﮭﺎز اﻟﻘﯾﺎس وﺟود ْ‬‫ﻧظم ﻏﯾر ﻣﻧﺗظم أﺛﻧﺎء ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‪ ،‬ﺳﯾظﮭر اﻟﻣؤﺷر ‪ IHB/AFib‬ﻋﻠﻰ‬‫اﻟﺷﺎﺷﺔ ﻣﻊ ﻗﯾم اﻟﻘﯾﺎس‪.‬‬‫‪ IHB/AFib‬ھذا ﺑﺻورة ﻣﺗﻛررة‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬ﻧﺣن ﻧوﺻﻲ ﺑﺎﻻﺗﺻﺎل ﺑﺎﻟطﺑﯾب إذا ظﮭر ﻟك ﻣؤﺷر‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪(AFib‬؟‬‫ﻧظرا ﻟﻺﺷﺎرات اﻟﻛﮭرﺑﺎﺋﯾﺔ اﻟواﻗﻌﺔ ﻓﻲ اﻟﻘﻠب وﯾﻧﺷر اﻟدم ﻓﻲ ﺟﻣﯾﻊ أﻧﺣﺎء اﻟﺟﺳم‪ .‬ﯾﺣدث‬‫ﯾﻧﻘﺑض اﻟﻘﻠب‬
‫ً‬
‫ُ‬
‫ُ‬
‫اﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪ (AFib‬ﻋﻧدﻣﺎ ﺗﺻﺑﺢ اﻹﺷﺎرة اﻟﻛﮭرﺑﺎﺋﯾﺔ ﻓﻲ اﻷذﯾن ﻣﺷوﺷﺔ‪ ،‬ﻣﺎ ﯾؤدي إﻟﻰ اﺿطراﺑﺎت‬‫ﻓﻲ اﻟﻔواﺻل اﻟزﻣﻧﯾﺔ ﺑﯾن اﻟﺿرﺑﺎت‪ .‬وﯾﻣﻛن أن ﯾﺳﺑب اﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪ (AFib‬رﻛود اﻟدم ﻓﻲ اﻟﻘﻠب‪ ،‬ﻣﺎ‬‫اﻟﺗﻌرض ﻟﻧوﺑﺔ ﻗﻠﺑﯾﺔ‪.‬‬‫ﯾﺷ ّﻛل ﺑﺳﮭوﻟﺔ ﺟﻠطﺎت دﻣوﯾﺔ ﺗﺗﺳﺑب ﻻﺣﻘًﺎ ﻓﻲ‬‫ّ‬‫‪%IHB/AFib‬‬‫ﯾظﮭر اﻟرﻣز ‪ %IHB/AFib‬ﻋﻠﻰ اﻟﺷﺎﺷﺔ ﻋﻧد ﺗﻛرار اﻛﺗﺷﺎف وﺟود ﺿرﺑﺎت ﻗﻠب ﻏﯾر ﻣﻧﺗظﻣﺔ أو رﺟﻔﺎن‬‫اُذﯾﻧﻲ‪.‬‬‫ﯾﻣﻛن أن ﯾظﮭر رﻣز ‪ IHB/AFib‬ﻟﯾس ﻓﻘط ﻋﻧد اﻛﺗﺷﺎف دﻗﺎت ﻗﻠب ﻏﯾر ﻣﻧﺗظﻣﺔ ﻓﺣﺳب ﺑل اﯾﺿﺎ ً ﻋﻧد‬‫ﺣدوث ﺿوﺿﺎء ﻛﺣرﻛﺔ اﻟﺟﺳم أﺛﻧﺎء اﻟﻘﯾﺎس‪ .‬ﻟذﻟك‪ ،‬ﻓﻧﺣن ﻧوﺻﻲ اﻻﺗﺻﺎل ﺑﺎﻟطﺑﯾب إذا ﻛﺎن‬‫ﻣﻌدل ‪ %IHB/AFib‬ﻣرﺗﻔﻌًﺎ‪.‬‬‫ﻋدد ﺣﺎﻻت ‪IHB/AFib‬‬‫اﻟ ُﻣﻛﺗﺷﻔﺔ ﻓﻲ اﻟذاﻛرة‬‫]‪x 100 [%‬‬‫= ‪%IHB/AFib‬‬‫اﻟرﻗم اﻟﻛﻠﻲ‬‫ﯾﺗم ﻋرض ‪ %IHB/AFib‬ﻋﻠﻰ اﻟﺷﺎﺷﺔ ﻋﻧد ﻋرض اﻟﻘﯾم اﻟﻣﺗوﺳطﺔ‪.‬‬‫ﻋرض ‪%IHB/AFib:‬‬‫)راﺟﻊ "‪ .2‬ﻛﯾﻔﯾﺔ ﻣراﺟﻌﺔ ﺑﯾﺎﻧﺎت اﻟذاﻛرة" ﻓﻲ "وﺿﻊ اﻟﺗﺷﻐﯾل"(‬‫ﻻ ﯾﺗم ﻋرض ‪ %IHB/AFib‬ﻋﻧدﻣﺎ ﯾﻛون رﻗم اﻟذاﻛرة ﺳﺗﺔ أو أﻗل‪.‬‬‫ﻋرض اﻟﻘﯾم اﻟﻣﺗوﺳطﺔ‬‫اﻟﻣﺳﺗوى ‪0‬‬‫‪%IHB/AFib=0‬‬‫اﻟﻣﺳﺗوى ‪1‬‬‫‪%IHB/AFib=1 - 9‬‬‫اﻟﻣﺳﺗوى ‪2‬‬‫‪%IHB/AFib=10 - 24‬‬‫ﻻ ﯾُﻌرض ﺷﻲء‬‫اﻟﻌرﺑﯾﺔ ‪16‬‬‫اﻟﻣﺳﺗوى ‪3‬‬‫‪%IHB/AFib=25 - 100‬‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‬‫ﯾراﻗب اﻟﻣؤﺷر ﻣدى اﻟﺗﻘدّم ﻓﻲ ﻋﻣﻠﯾﺔ ﻗﯾﺎس اﻟﺿﻐط‪.‬‬‫ﺟﺎر ﻗﯾﺎس اﻟﺿﻐط‬‫ٍ‬‫ﺟﺎر ﺗﻔرﯾﻎ اﻟﮭواء‬‫ٍ‬‫ﻧ ْﻔﺦ اﻟﮭواء ﻗﯾد اﻟﺗﻧﻔﯾذ‬‫اﻛﺗﻣل ﻧ ْﻔﺦ اﻟﮭواء‬‫ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﺗﺗواﻓﻖ ﻛل ﺷرﯾﺣﺔ ﻣن ﺷراﺋﺢ ﻣؤﺷر اﻟﺷرﯾط ﻣﻊ ﺗﺻﻧﯾف‬‫ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﻣوﺿﺢ ﻓﻲ‬‫اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم اﻟﺷدﯾد‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم اﻟﻣﺗوﺳط‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم اﻟﺧﻔﯾف‬‫ﻧطﺎق ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫طﺑﯾﻌﻲ‬‫ﻣﺛﺎﻟﻲ‬‫ﯾﻌرض اﻟﻣؤﺷر ﺷرﯾﺣﺔ ﺗﺑﻌًﺎ ﻟﻠﺑﯾﺎﻧﺎت اﻟﺣﺎﻟﯾﺔ وﺑﻣﺎ ﯾﺗواﻓﻖ ‪:‬‬
‫ﻣﻊ ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‪.‬‬‫ﻣﺛﺎل‪:‬‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم اﻟﻣﺗوﺳط‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم اﻟﺧﻔﯾف‬‫ﻧطﺎق ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫ﻧُﺑذة ﻋن ﺿﻐط اﻟدم‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺿﻐط اﻟدم؟‬‫ﺿﻐط اﻟدم ھو ﻗوة دﻓﻊ اﻟدم ﻟﺟدران اﻟﺷراﯾﯾن‪ .‬وﻋﻧدﻣﺎ ﯾﻧﻘﺑض اﻟدم‪ ،‬ﯾﺣدث اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪ .‬وﻋﻧدﻣﺎ ﯾﻧﺑﺳط‬‫اﻟدم‪ ،‬ﯾﺣدث اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ .‬وﯾﻛون اﻟﻣﻠﻠﯾﻣﺗر اﻟزﺋﺑﻘﻲ )‪ (mmHg‬ھو وﺣدة ﻗﯾﺎس ﺿﻐط اﻟدم‪ .‬وﯾُﻣﺛل‬‫اﻟﺿﻐط اﻷﺳﺎﺳﻲ ﻣﻘﯾﺎس ﺿﻐط اﻟدم اﻟطﺑﯾﻌﻲ ﻟﻠﻔرد‪ ،‬وھو ّأول ﻣﺎ ﯾُﻘﺎس ﻓﻲ اﻟﺻﺑﺎح ﺣﯾث ﯾﻛون اﻟﻔرد ﻣﺳﺗرﺧﯾًﺎ‬‫وﻗﺑل ﺗﻧﺎوﻟﮫ اﻟطﻌﺎم‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪17‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎرﺗﻔﺎع ﺿﻐط اﻟدم وﻛﯾف ﯾﻣﻛن اﻟﺗﺣﻛم ﻓﯾﮫ؟‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم ھو ﺣﺎﻟﺔ ﻣن ﺿﻐط اﻟدم اﻟﺷرﯾﺎﻧﻲ اﻟﻣرﺗﻔﻊ ﻏﯾر اﻟطﺑﯾﻌﻲ‪ ،‬اﻟذي ﯾﻣﻛن أن ﯾﺗﺳﺑب‪ ،‬ﻓﻲ ﺣﺎﻟﺔ‬‫اﻟﺗﻌرض ﻟﻣﺷﻛﻼت ﺻﺣﯾﺔ‪ ،‬ﻣن ﺑﯾﻧﮭﺎ اﻟﻧوﺑﺔ اﻟﻘﻠﺑﯾﺔ‪ .‬وﯾﻣﻛن اﻟﺗﺣﻛم ﻓﻲ ارﺗﻔﺎع ﺿﻐط اﻟدم‬‫ﻋدم اﻟﺗﺣﻛم ﻓﯾﮫ‪ ،‬ﻓﻲ‬‫ّ‬‫ﻋن طرﯾﻖ ﺗﻐﯾﯾر ﻧﻣط اﻟﺣﯾﺎة‪ ،‬وﺗﻔﺎدي ﺣﺎﻻت اﻹﺟﮭﺎد‪ ،‬وﺑﺗﻧﺎول اﻟﻌﻼج ﺗﺣت اﻹﺷراف اﻟطﺑﻲ‪.‬‬‫ﻟﻠوﻗﺎﯾﺔ ﻣن ارﺗﻔﺎع ﺿﻐط اﻟدم أو اﻟﺗﺣﻛم ﻓﯾﮫ‪:‬‬‫ﻣﺎرس اﻟرﯾﺎﺿﺔ ﺑﺎﻧﺗظﺎم‬‫ﺗﺟﻧب اﻟﺗدﺧﯾن‬‫اﺧﺿﻊ ﻟﻔﺣوﺻﺎت طﺑﯾﺔ ﻣﻧﺗظﻣﺔ‬‫ّﻗﻠل ﻣن ﺗﻧﺎول اﻷﻣﻼح واﻟدھون‬‫ﺣﺎﻓظ ﻋﻠﻰ وزﻧك اﻟﻣﺛﺎﻟﻲ‬‫ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟‬‫إن ﻗﯾﺎس ﺿﻐط اﻟدم ﻓﻲ ﻋﯾﺎدة أو ﻋﻧد اﻟطﺑﯾب ﯾﻣﻛن أن ﯾﺛﯾر اﻟﻣﺧﺎوف وﯾُﻌطﻲ ﻗراءة ﻣرﺗﻔﻌﺔ‪ ،‬ﺑﻣﻘدار ‪ 25‬إﻟﻰ‬‫‪ 30‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ أﻋﻠﻰ ﻣن اﻟﻘﯾﺎس ﻓﻲ اﻟﻣﻧزل‪ .‬ﯾﻘﻠل اﻟﻘﯾﺎس ﻓﻲ اﻟﻣﻧزل ﻣن اﻟﺗﺄﺛﯾرات اﻟﺧﺎرﺟﯾﺔ ﻋﻠﻰ ﻗراءات‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬وﯾﻣﺛل إﺿﺎﻓﺔ إﻟﻰ ﻗراءات اﻟطﺑﯾب‪ ،‬وﯾﻘدم ﺳﺟ ً‬‫ﻼ ﻣن اﻟﻘراءات اﻟﻛﺎﻣﻠﺔ واﻷﻛﺛر دﻗﺔ ﺑﺎﻟﻧﺳﺑﺔ‬‫ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم ﻣن اﻟﺗﺻﻧﯾف ‪) 3‬ﺷدﯾد(‬‫وﺿﻌت ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻣﻌﺎﯾﯾر ﻟﺗﻘﯾﯾم‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ‪ ،‬دون اﻟﻧظر إﻟﻰ اﻟﻌﻣر‪،‬‬‫ﻛﻣﺎ ﯾﺗﺿﺢ ذﻟك ﻓﻲ اﻟرﺳم اﻟﺗﺧطﯾطﻲ‪.‬‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم ﻣن اﻟﺗﺻﻧﯾف ‪) 2‬ﺧﻔﯾف(‬‫ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى‬‫ﺿﻐط اﻟدم‬‫اﻟﻣﺎدة اﻟﻣرﺟﻌﯾﺔ؛ ‪Journal of Hypertension 1999, Vol 17 No.2‬‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم ﻣن اﻟﺗﺻﻧﯾف ‪) 1‬ﺧﻔﯾف(‬‫اﻟﻧطﺎق ﻣن ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫طﺑﯾﻌﻲ‬‫ﻣﺛﺎﻟﻲ‬‫ﺣﺎﻻت اﻻﺧﺗﻼف ﻓﻲ ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫ﯾﺧﺗﻠف ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟﻠﻔرد اﺧﺗﻼﻓًﺎ‬‫ﻧظرا‬‫ﻛﺑﯾرا ﻋﻠﻰ ﻣدار اﻟﯾوم واﻟﻣوﺳم‪ .‬وﯾﻣﻛن أن ﯾﺗراوح ﻣﻘدار اﻻﺧﺗﻼف ﺑﯾن ‪ 30‬و‪ 50‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ ً‬‫ً‬‫ﻟﻠظروف اﻟﻣﺧﺗﻠﻔﺔ طوال اﻟﯾوم‪ .‬وﯾﻣﻛن أن ﺗﻛون اﻻﺧﺗﻼﻓﺎت ﻟدى اﻷﻓراد اﻟذﯾن ﯾﻌﺎﻧون ﻣن ارﺗﻔﺎع ﺿﻐط‬‫اﻟدم ﻣﻠﺣوظﺔ ﺑﻘدر أﻛﺑر‪ .‬وﻋﺎدة ﻣﺎ ﯾرﺗﻔﻊ ﻣﺳﺗوى ﺿﻐط اﻟدم أﺛﻧﺎء اﻟﻌﻣل أو اﻟﻠﻌب‪ ،‬وﯾﻧﺧﻔض إﻟﻰ أدﻧﻰ‬‫ﻛﺛﯾرا ﺣﯾﺎل اﻟﻧﺗﺎﺋﺞ اﻟﺗﻲ‬‫ﻣﺳﺗوﯾﺎﺗﮫ أﺛﻧﺎء اﻟﻧوم‪ .‬ﻟذﻟك‪ ،‬ﻻ ﺗﻘﻠﻖ‬‫ً‬‫اﻟﺗﻐﯾّرات اﻟﻧﻣطﯾﺔ ﻋﻠﻰ ﻣدار اﻟﯾوم‬‫)اﻟﻘﯾﺎس ﻛل ﺧﻣس دﻗﺎﺋﻖ(‬‫ﺗظﮭر ﻣن ﻋﻣﻠﯾﺔ ﻗﯾﺎس واﺣدة‪.‬‬‫اﻧﻘﺑﺎﺿﻲ‬‫أﺟر ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ ﻛل ﯾوم ﺑﺎﺗﺑﺎع اﻹﺟراء‬‫اﻟﻣوﺿّﺢ ﻓﻲ ھذا اﻟدﻟﯾل ﻟﻣﻌرﻓﺔ اﻟﻣﻌدل اﻟطﺑﯾﻌﻲ ﻟﻣﺳﺗوى‬‫ﺿﻐط اﻟدم ﻟدﯾك‪ .‬وﺗﻘدّم اﻟﻘراءات اﻟﻣﻧﺗظﻣﺔ ﺳﺟﻼً أﻛﺛر‬‫ﺷﻣوﻟﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪ .‬اﺣرص ﻋﻠﻰ ﺗوﺿﯾﺢ اﻟوﻗت‬‫واﻟﺗﺎرﯾﺦ ﻋﻧد ﺗﺳﺟﯾل ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪ .‬اﺳﺗﺷر‬‫اﻧﺑﺳﺎطﻲ‬‫ﯾﻧﺎم‬‫اﻟطﺑﯾب ﻟﯾﻔﺳّر ﻟك ﺑﯾﺎﻧﺎت ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫اﻟوﻗت‬‫ﺿﻐط اﻟدم اﻻﻧﻘﺑﺎﺿﻲ‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫اﻟﻌرﺑﯾﺔ ‪18‬‬‫اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ‬‫اﻟﻣﺷﻛﻠﺔ‬‫ﻻ ﯾظﮭر ﺷﻲء ﻋﻠﻰ‬‫اﻟﺷﺎﺷﺔ‪ ،‬ﺣﺗﻰ ﻋﻧد‬‫ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫اﻟﺳﺑب اﻟﻣﺣﺗﻣل‬‫ﻧﻔدت طﺎﻗﺔ اﻟﺑطﺎرﯾﺎت‪.‬‬‫أطراف اﻟﺑطﺎرﯾﺎت ﻟﯾﺳت ﻓﻲ‬‫ﻣواﺿﻌﮭﺎ اﻟﺻﺣﯾﺣﺔ‪.‬‬‫ﺟﮭد اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔض ﻟﻠﻐﺎﯾﺔ‪.‬‬‫ﯾوﻣض )رﻣز اﻟﺑطﺎرﯾﺔ‬‫ﻻ ﯾﻧﺗﻔﺦ اﻟرﺑﺎط ﺑﺎﻟﮭواء‪.‬‬‫ﻣﻧﺧﻔﺿﺔ(‪ .‬إذا ﻧﻔدت طﺎﻗﺔ اﻟﺑطﺎرﯾﺎت‬‫ﺑﺎﻟﻛﺎﻣل‪ ،‬ﻓﻼ ﯾظﮭر اﻟرﻣز‪.‬‬‫ﻟم ﯾُﻠف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﻣﻼﺋﻣﺔ‪.‬‬‫ﺣرﻛت ذراﻋك أو ﺟﺳدك أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ّ‬‫اﻟﺟﮭﺎز ﻻ ﯾﻘﯾس‪.‬‬‫اﻟﻘراءات ﻣرﺗﻔﻌﺔ أو‬‫ﻣﻧﺧﻔﺿﺔ ﻟﻠﻐﺎﯾﺔ‪.‬‬‫أﺧرى‬‫ﻣوﺿﻊ اﻟرﺑﺎط ﻏﯾر ﺻﺣﯾﺢ‪.‬‬‫اﻟﻘﯾﻣﺔ ﻣﺧﺗﻠﻔﺔ ﻋن اﻟﻘﯾﺎس ﻓﻲ اﻟﻌﯾﺎدة‬‫أو ﻟدى اﻟطﺑﯾب‪.‬‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪.‬‬‫أ ِﻋد ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت ﺑﺣﯾث ﺗﻛون اﻷطراف‬‫اﻟﺳﺎﻟﺑﺔ واﻟﻣوﺟﺑﺔ ﻣﺗطﺎﺑﻘﺔ ﻣﻊ ﺗﻠك اﻟﻣوﺿﺣﺔ‬‫ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت‪.‬‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪.‬‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫اﺣرص ﻋﻠﻰ أن ﺗظل ﺛﺎﺑﺗًﺎ وھﺎدﺋًﺎ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﺟﻠس ﻓﻲ ﺣﺎﻟﺔ ﺛﺑﺎت وھدوء‪.‬‬‫ﺿﻊ ذراﻋك ﻋﻠﻰ طﺎوﻟﺔ ﺑﺣﯾث ﺗﻛون راﺣﺔ ﯾدك‬‫ﻣواﺟﮭﺔ ﻷﻋﻠﻰ واﻟرﺑﺎط ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‪.‬‬‫ً‬‫إذا ﻛﺎن ﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟدﯾك ﺿﻌﯾﻔﺎ ﺟدًا أو‬‫ﻏﯾر ﻣﻧﺗظم‪ ،‬ﻓﻘد ﯾﺟد اﻟﺟﮭﺎز ﺻﻌوﺑﺔ ﻓﻲ ﺗﺣدﯾد‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪.‬‬‫راﺟﻊ "ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟"‪.‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت‪ .‬ر ّﻛﺑﮭﺎ ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪،‬‬‫وﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬إذا ﻟم ﺗُﺣل اﻟﻣﺷﻛﻼت ﺑﺎﺗﺑﺎع اﻹﺟراءات اﻟﻣوﺿﺣﺔ أﻋﻼه‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل‪ .‬ﻻ ﺗﺣﺎول ﻓﺗﺢ أو إﺻﻼح‬‫ﻧظرا ﻷن أي ﻣﺣﺎوﻟﺔ ﻟﻠﻘﯾﺎم ﺑذﻟك ﺳﺗؤدي إﻟﻰ إﻟﻐﺎء اﻟﺿﻣﺎن‪.‬‬‫ھذا اﻟﻣﻧﺗﺢ‪ً ،‬‬‫اﻟﻌرﺑﯾﺔ ‪19‬‬‫اﻟﺻﯾﺎﻧﺔ‬‫ﻣﻛوﻧﺎت ﻛﮭرﺑﺎﺋﯾﺔ ﺷدﯾدة اﻟﺣﺳﺎﺳﯾﺔ ووﺣدة ھواء ﻣﻌﻘدة ﯾﻣﻛن أن ﺗﺗﻠف‪ .‬وإذا ﺗﻌ ّذر‬‫ﻻ ﺗﻔﺗﺢ اﻟﺟﮭﺎز‪ .‬ﺗُﺳﺗﺧدم ّ‬‫ﻋﻠﯾك ﺣل اﻟﻣﺷﻛﻠﺔ ﺑﺎﺗﺑﺎع إرﺷﺎدات اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل اﻟﻣﻌﺗﻣد ﻓﻲ ﻣﻧطﻘﺗك أو‬‫ﺳﯾزود ﻗﺳم ﺧدﻣﺔ اﻟﻌﻣﻼء اﻟﺗﺎﺑﻊ ﻟﺷرﻛﺔ ‪ A&D‬اﻟﻣﻌﻠوﻣﺎت اﻟﻔﻧﯾﺔ وﻗطﻊ اﻟﻐﯾﺎر واﻟوﺣدات‬‫ﻗﺳم ﺧدﻣﺔ اﻟﻌﻣﻼء‪.‬‬‫ّ‬‫ﻟﻠوﻛﻼء اﻟﻣﻌﺗﻣدﯾن‪.‬‬‫ﺻﻧّﻊ ﻟﯾﺗﻣﺗﻊ ﺑﻔﺗرة ﺗﺷﻐﯾل طوﯾﻠﺔ‪ .‬وﻣﻊ ذﻟك‪ ،‬ﻓﺈﻧﮫ ﯾوﺻﻰ ﺑﺷﻛل ﻋﺎم ﺑﻔﺣص اﻟﺟﮭﺎز ﻛل‬‫ﺻﻣم اﻟﺟﮭﺎز و ُ‬‫ُ‬‫ﻋﺎﻣﯾن ﻟﺿﻣﺎن اﻟﺗﺷﻐﯾل اﻟﺻﺣﯾﺢ ﻟﮫ ودﻗﺔ أداﺋﮫ‪ .‬ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑﺎﻟوﻛﯾل اﻟﻣﻌﺗﻣد ﻓﻲ ﻣﻧطﻘﺗك أو ﺷرﻛﺔ‬‫‪ A&D‬ﻹﺟراء اﻟﺻﯾﺎﻧﺔ‪.‬‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻔﻧﯾﺔ‬‫اﻟﻧوع‬‫طرﯾﻘﺔ اﻟﻘﯾﺎس‬‫ﻧطﺎق اﻟﻘﯾﺎس‬‫دﻗﺔ اﻟﻘﯾﺎس‬‫ﻣﺻدر اﻟطﺎﻗﺔ‬‫ﻋدد ﻣرات اﻟﻘﯾﺎس‬‫اﻟﺗﺻﻧﯾف‬‫اﻻﺧﺗﺑﺎر اﻟﺳرﯾري‬‫اﻟﺗواﻓﻖ اﻟﻛﮭروﻣﻐﻧﺎطﯾﺳﻲ‬‫اﻟذاﻛرة‬‫ظروف اﻟﺗﺷﻐﯾل‬‫ظروف اﻟﻧﻘل ‪ /‬اﻟﺗﺧزﯾن‬‫اﻷﺑﻌﺎد‬‫اﻟوزن‬‫اﻟﺣﻣﺎﯾﺔ ﻣن اﻟﻣواد اﻟدﺧﯾﻠﺔ‬‫‪UA-767S-W‬‬‫اﻟﻘﯾﺎس ﺑﺎﻟذﺑذﺑﺎت‬‫‪ 299 - 0‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ‬‫اﻟﺿﻐط‪:‬‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪ 279 - 60 :‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ‬‫اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ 200 - 40 :‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ‬‫‪ 180 - 40‬ﺿرﺑﺔ ‪ /‬اﻟدﻗﯾﻘﺔ‬‫اﻟﻧﺑض‪:‬‬‫‪ 3±‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ‬‫اﻟﺿﻐط‪:‬‬‫‪%5±‬‬‫اﻟﻧﺑﺿﺎت‪:‬‬‫‪ 4‬ﺑطﺎرﯾﺎت ﺑﺟﮭد ‪ 1.5‬ﻓوﻟت )ﻣﻘﺎس ‪ R6P‬أو ‪ LR6‬أو ‪ (AA‬أو‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد )‪) (TB-233C‬ﻏﯾر ﻣرﻓﻖ(‬‫ّ‬‫اﻟﺑطﺎرﯾﺔ ﻣﻘﺎس ‪) LR6‬ﺑطﺎرﯾﺎت ﻗﻠوﯾﺔ(‬‫‪ 700‬ﻣرة ﻗﯾﺎس ﺗﻘرﯾﺑًﺎ‬‫اﻟﺑطﺎرﯾﺔ ﻣﻘﺎس ‪) R6P‬ﺑطﺎرﯾﺎت ﻣﻧﺟﻧﯾز(‬‫‪ 200‬ﻣرة ﻗﯾﺎس ﺗﻘرﯾﺑًﺎ‬‫ﻋﻧد ﻗﯾﻣﺔ ﺿﻐط ‪ 180‬ﻣﻠﻠﯾﻣﺗر زﺋﺑﻘﻲ‪ ،‬ودرﺟﺔ ﺣرارة اﻟﻐرﻓﺔ ‪ 23‬درﺟﺔ‬‫ﻣﺋوﯾﺔ‪.‬‬‫ﺟﮭﺎز طﺑﻲ ﻛﮭرﺑﺎﺋﻲ ﺑﻣﺻدر طﺎﻗﺔ داﺧﻠﻲ )ﯾﻌﻣل ﺑﺑطﺎرﯾﺎت( ‪ /‬اﻟﻔﺋﺔ ‪II‬‬‫ﺑﻣﺣول اﻟﺗﯾﺎر(‬‫)ﯾﻌﻣل‬‫ّ‬‫وﺿﻊ اﻟﺗﺷﻐﯾل اﻟﻣﺳﺗﻣر‬‫وﻓﻘًﺎ ﻟﺷﮭﺎدة ‪ISO81060-2：2013‬‬‫ﻋﺎ؛ ﻟﺗﺣدﯾد ﻣﺳﺗوى ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ‬‫ﺗم اﺳﺗﺧدام ‪ K5‬ﻓﻲ ‪ 85‬ﻣوﺿو ً‬‫ﻓﻲ دراﺳﺔ ﺗﺣﻘﱡﻖ ﺳرﯾرﯾﺔ‪.‬‬‫اﻟﺗوﺟﯾﮫ ‪IEC 60601-1-2: 2014‬‬‫آﺧر ‪ 60‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس‬‫‪ +10‬إﻟﻰ ‪ +40‬درﺟﺔ ﻣﺋوﯾﺔ ‪ 15 /‬إﻟﻰ ‪ %85‬اﻟرطوﺑﺔ اﻟﻧﺳﺑﯾﺔ ‪800 /‬‬‫إﻟﻰ ‪ 1,060‬ھﻛﺗوﺑﺎﺳﻛﺎل‬‫‪ -20‬إﻟﻰ ‪ +60‬درﺟﺔ ﻣﺋوﯾﺔ ‪ 10 /‬إﻟﻰ ‪ %95‬اﻟرطوﺑﺔ اﻟﻧﺳﺑﯾﺔ ‪700 /‬‬‫إﻟﻰ ‪ 1,060‬ھﻛﺗوﺑﺎﺳﻛﺎل‬‫‪] 140‬ﻋرض[ × ‪] 60‬ارﺗﻔﺎع[ × ‪] 105‬ﻋﻣﻖ[ ﻣم ﺗﻘرﯾﺑًﺎ‬‫‪ 245‬ﺟرام ﺗﻘرﯾﺑًﺎ‪ ،‬ﺑدون اﻟﺑطﺎرﯾﺎت‬‫‪IP20‬‬‫اﻟﺟﮭﺎز‪:‬‬‫اﻟﻌرﺑﯾﺔ ‪20‬‬‫اﻟﻧوع ‪BF‬‬‫اﻟﺟزء اﻟﻣﻠﻔوف‬‫اﻟرﺑﺎط‬‫ﻣدة اﻟﺻﻼﺣﯾﺔ‬‫‪ 5‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫اﻟﺟﮭﺎز‪:‬‬‫ﺳﻧﺗﺎن )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫اﻟرﺑﺎط‪:‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪ 5 :‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻣﻠﺣﻖ ّ‬‫ﻣﺻدرا ﻟﻺﻣداد ﺑﺎﻟطﺎﻗﺔ ﻓﻲ اﻟﻣﻧزل‪ .‬ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑوﻛﯾل ‪ A&D‬اﻟﻣﺣﻠﻲ‬‫اﻟﻣﺣول ﺑﺎﻟﺟﮭﺎز ﻟﯾﻛون‬‫ﯾﺗﺻل‬‫ً‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد أو اﺳﺗﺑداﻟﮫ ﺑﺷﻛل دوري‪.‬‬‫ﻟﺷراﺋﮫ‪ .‬ﯾﻛون ﺿرورﯾًﺎ ﻓﺣص ّ‬‫ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑوﻛﯾل ‪ A&D‬اﻟﻣﺣﻠﻲ ﻟﺷراﺋﮫ‪.‬‬‫‪TB-233C‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد أو اﺳﺗﺑداﻟﮫ ﺑﺷﻛل دوري‪.‬‬‫ﯾﻛون ﺿرورﯾًﺎ ﻓﺣص ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ ّ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫اﻟرﻣوز‬‫ﻟﻼﺳﺗﺧدام اﻟﻣﻧزﻟﻲ ﻓﻘط‬‫اﻟﺟﮭﺎز ﻣن اﻟﻔﺋﺔ ‪II‬‬‫اﻟﻣﺻﮭر اﻟﺣراري‬‫اﻟﻣﺻﮭر‬‫ﻣﻠﺻﻖ اﻟﺗوﺟﯾﮫ ‪ EC‬ﻟﻸﺟﮭزة‬‫ﻣﻠﺻﻖ ﺷﮭﺎدة ‪ EAC‬ﻟﻸﺟﮭزة‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟﻘطﺑﯾﺔ ﻟﻘﺎﺑس ّ‬‫اﻟﻣﻠﺣﻘﺎت ﺗُﺑﺎع ﺑﺷﻛل ﻣﻧﻔﺻل‬‫ﺣﺟم اﻟرﺑﺎط‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫ا ﻟرﺑﺎط‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬‫‪CUF-F-LA‬‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬‫‪CUF-I‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪CUF-F-A‬‬‫ﺣﺟم اﻟذراع‪ :‬اﻟﻣﺣﯾط ﻋﻧد ﻋﺿﻠﺔ اﻟذراع‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ّ‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫‪TB-233C‬‬‫ﺣﺟم اﻟذراع‬‫‪ 31‬ﺳم إﻟﻰ ‪ 45‬ﺳم‬‫‪ 22‬ﺳم إﻟﻰ ‪ 42‬ﺳم‬‫‪ 22‬ﺳم إﻟﻰ ‪ 32‬ﺳم‬‫اﻟﻘﺎﺑس )ﻧوع اﻟﻣﺄﺧذ(‬‫اﻟﻧوع ‪C‬‬

‫ﻣﻼﺣظﺔ‪ :‬ﺗﺧﺿﻊ اﻟﻣواﺻﻔﺎت ﻟﻠﺗﻐﯾﯾر دون إﺷﻌﺎر ﻣﺳﺑﻖ‪.‬‬‫ﺗﺻﻧﯾف ‪ IP‬ھو درﺟﺎت اﻟﺣﻣﺎﯾﺔ اﻟﺗﻲ ﯾﺗﻣﺗﻊ ﺑﮭﺎ اﻟﺟﮭﺎز وﻓﻘًﺎ ﻟﻠﺗوﺟﯾﮫ ‪ .IEC 60529‬ھذا اﻟﺟﮭﺎز‬‫ﻣﺣﻣﻲ ﺿد اﻟﻣواد اﻟﻐرﯾﺑﺔ اﻟﺻﻠﺑﺔ اﻟﺗﻲ ﯾﺑﻠﻎ ﻗطرھﺎ ‪ 12‬ﻣم أو أﻛﺑر‪ ،‬ﻣﺛل اﻷﺻﺎﺑﻊ‪ .‬ھذا اﻟﺟﮭﺎز ﻏﯾر‬‫ﻣﺣﻣﻲ ﺿد اﻟﻣﺎء‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪21‬‬‫ﻣذﻛرة‬‫اﻟﻌرﺑﯾﺔ ‪22‬‬‫ﻓﮭرﺳت ﻣﻧدرﺟﺎت‬‫ﻣﺷﺗرﯾﺎن ﮔراﻣﯽ ‪2 .......................................................................................‬‬‫ﺗذﮐرات اوﻟﯾﮫ‪2 ...........................................................................................‬‬‫اﻗداﻣﺎت اﺣﺗﯾﺎطﯽ ‪2 ......................................................................................‬‬‫ﻣﻌرﻓﯽ ﻗطﻌﺎت ‪5 .........................................................................................‬‬‫ﻧﻣﺎدھﺎ‪6 ... ................................................................................................‬‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت ‪8 ..........................................................................................‬‬‫‪ .1‬اﻧدازهﮔﯾری ﻧرﻣﺎل ‪8 ........................................... ................................‬‬‫‪ .2‬ﻓراﺧواﻧﯽ اطﻼﻋﺎت ‪8 .......................................... ................................‬‬‫‪ .۳‬ﺣذف ﮐﻠﯾﮫ اطﻼﻋﺎت ذﺧﯾرهﺷده در ﺣﺎﻓظﮫ ‪8 .................................................‬‬‫‪ .۴‬اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب ‪8 ......................................................‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش ‪9 ...............................................................................‬‬‫ﻧﺻب ‪ /‬ﺗﻌوﯾض ﺑﺎطریھﺎ ‪9 ...................................... ................................‬‬‫ﻣﺗﺻل ﮐردن ﺷﯾﻠﻧﮓ ھوا ‪10 ...................................... ................................‬‬‫ﻣﺗﺻل ﮐردن آداﭘﺗور ‪10 ..................................... ................................ AC‬‬‫اﻧﺗﺧﺎب ﺳﺎﯾز ﻣﻧﺎﺳب ﺑﺎزوﺑﻧد ‪11 .................................. ................................‬‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد ‪11 .............................................. ................................‬‬‫ﻧﺣوه اﻧﺟﺎم اﻧدازهﮔﯾری دﻗﯾﻖ ‪13 .................................. ................................‬‬‫اﻧدازهﮔﯾری ‪13 ..................................................... ................................‬‬‫ﭘس از اﻧدازهﮔﯾری ‪13 ............................................. ................................‬‬‫اﻧدازهﮔﯾری‪14 ...........................................................................................‬‬‫اﻧدازهﮔﯾری ﻧرﻣﺎل ‪14 ............................................. ................................‬‬‫اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب ‪15 ........................................................‬‬‫ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ ‪15 ................................. ................................‬‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ ‪16 .........................................................................‬‬‫ﺷﺎﺧص ‪ IHB/AFib‬ﭼﯾﺳت؟ ‪17 ......................................................................‬‬‫‪ AFib‬ﭼﯾﺳت؟ ‪17 ......................................................................................‬‬‫‪17 ...................................................................................... %IHB/AFib‬‬‫ﺷﺎﺧص ﺑﺎر ﻓﺷﺎر ‪18 ....................................................................................‬‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪18 ..............................................(WHO‬‬‫در ﻣورد ﻓﺷﺎر ﺧون‪18 .................................................................................‬‬‫ﻓﺷﺎر ﺧون ﭼﯾﺳت؟ ‪18 ............................................. ................................‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﭼﯾﺳت و ﭼﮕوﻧﮫ ﮐﻧﺗرل ﻣﯽﺷود؟ ‪19 ............................................‬‬‫ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ اﻧدازهﮔﯾری ﮐﻧم؟ ‪19 ................................................‬‬‫طﺑﻘﮫﺑﻧدی ﻓﺷﺎر ﺧون ‪19 ................................. ................................ WHO‬‬‫ﻧوﺳﺎﻧﺎت ﻓﺷﺎر ﺧون ‪19 ............................................ ................................‬‬‫ﻋﯾبﯾﺎﺑﯽ ‪20 .............................................................................................‬‬‫ﻧﮕﮭداری ‪21 .............................................................................................‬‬‫اطﻼﻋﺎت ﻓﻧﯽ ‪21 ........................................................................................‬‬‫ﻓﺎرﺳﯽ ‪1‬‬‫ﻣﺷﺗرﯾﺎن ﮔراﻣﯽ‬‫ﺑﺎﺑت ﺧرﯾد ﺟدﯾدﺗرﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ‪ A&D‬ﮐﮫ ﯾﮑﯽ از ﭘﯾﺷرﻓﺗﮫﺗرﯾن دﺳﺗﮕﺎهھﺎی ﭘﺎﯾش ﻣوﺟود‬‫در ﺑﺎزار اﺳت‪ ،‬ﺑﮫ ﺷﻣﺎ ﺗﺑرﯾﮏ ﻣﯽﮔوﯾﯾم‪ .‬اﯾن دﺳﺗﮕﺎه ﮐﮫ ﺑرای اﺳﺗﻔﺎده آﺳﺎن و در ﻋﯾن ﺣﺎل دﻗﯾﻖ طراﺣﯽ‬‫ﺷده اﺳت‪ ،‬ﺑررﺳﯽ روزاﻧﮫ ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﺳﮭﯾل ﻣﯽﮐﻧد‪.‬‬‫ﺗوﺻﯾﮫ ﻣﯽﺷود ﭘﯾش از آﻧﮑﮫ ﺑرای اوﻟﯾن ﺑﺎر از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬ﮐﺗﺎﺑﭼﮫ راھﻧﻣﺎ را ﺑﮫدﻗت‬‫ﻣطﺎﻟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﺗذﮐرات اوﻟﯾﮫ‬‫اﯾن دﺳﺗﮕﺎه ﺑﺎ دﺳﺗوراﻟﻌﻣل ‪ 93/42 EEC‬ﺑرای ﻣﺣﺻوﻻت ﭘزﺷﮑﯽ در اروﭘﺎ ﻣطﺎﺑﻘت دارد‪ .‬ﻧﺷﺎن‬‫ﺷﺎھدی ﺑر اﯾن ﻣدﻋﯽ اﺳت‪ :0123) .‬ﺷﻣﺎره ارﺟﺎع ﺑﮫ ﺑﺧش اﻋﻼم ﺷده ﻣرﺑوطﮫ(‬‫اﻧطﺑﺎق‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﺳﺗﻔﺎده ﺑزرﮔﺳﺎﻻن طراﺣﯽ ﺷده اﺳت‪ ،‬ﻧﮫ ﻧوزادان ﯾﺎ اطﻔﺎل‪.‬‬‫ﻣﺣﯾط ﮐﺎرﺑرد‪ .‬اﯾن دﺳﺗﮕﺎه ﺑﮫﮔوﻧﮫای طراﺣﯽ ﺷده اﺳت ﮐﮫ ﺷﻣﺎ ﺑﺗواﻧﯾد در ﻣﺣﯾط ﻣﻧزﻟﺗﺎن ﺷﺧﺻﺎ ً از آن‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون و ﺿرﺑﺎن ﻗﻠب اﻓراد ﺑرای ﺗﺷﺧﯾص ﺑﯾﻣﺎری طراﺣﯽ ﺷده اﺳت‪.‬‬‫اﻗداﻣﺎت اﺣﺗﯾﺎطﯽ‬‫در ﺳﺎﺧت اﯾن دﺳﺗﮕﺎه از اﺟزای دﻗﯾﻖ اﺳﺗﻔﺎده ﺷده اﺳت‪ .‬دﻣﺎ و رطوﺑت ﺷدﯾد‪ ،‬ﻧور ﻣﺳﺗﻘﯾم ﺧورﺷﯾد‪ ،‬ﺿرﺑﮫ‬‫ﯾﺎ ﮔرد و ﺧﺎک ﻣﯽﺗواﻧد ﺑﮫ آن آﺳﯾب ﺑرﺳﺎﻧد‪.‬‬‫ﺑﺎزوﺑﻧد و دﺳﺗﮕﺎه را ﺑﺎ ﯾﮏ ﭘﺎرﭼﮫ ﻧرم و ﺧﺷﮏ ﯾﺎ ﭘﺎرﭼﮫ ﻧﻣدار ﺷده ﺑﺎ آب ﯾﺎ ﺷوﯾﻧده ﺧﻧﺛﯽ ﺗﻣﯾز ﮐﻧﯾد‪.‬‬‫ھرﮔز از اﻟﮑل‪ ،‬ﺑﻧزﯾن‪ ،‬ﺗﯾﻧر ﯾﺎ دﯾﮕر ﻣواد ﺷﯾﻣﯾﺎﯾﯽ ﺧورﻧده ﺑرای ﺗﻣﯾز ﮐردن ﺑﺎزوﺑﻧد ﯾﺎ دﺳﺗﮕﺎه اﺳﺗﻔﺎده‬‫ﻧﮑﻧﯾد‪.‬‬‫از ﺧم ﮐردن زﯾﺎد ﺑﺎزوﺑﻧد ﯾﺎ ﭘﯾﭼﺎﻧدن ﺷﯾﻠﻧﮓ ﺑرای ﻣدت طوﻻﻧﯽ اﺟﺗﻧﺎب ﮐﻧﯾد‪ ،‬زﯾرا اﯾن ﮐﺎر ﻣوﺟب ﮐوﺗﺎه‬‫ﺷدن ﻋﻣر اﺟزای دﺳﺗﮕﺎه ﻣﯽﺷود‪.‬‬‫ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ ﺷﯾﻠﻧﮓ و ﮐﺎﺑل دﺳﺗﮕﺎه ﺑﮫ دور ﮔردن ﮐودﮐﺎن ﯾﺎ ﻧوزادان ﻧﭘﯾﭼد‪ ،‬زﯾرا ﻣﯽﺗواﻧد ﻣوﺟب ﺧﻔﮕﯽ‬‫آﻧﮭﺎ ﺷود‪.‬‬‫ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ در ﺣﯾن اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺷﯾﻠﻧﮓ ھوا دﭼﺎر ﭘﯾﭻﺧوردﮔﯽ ﻧﺷود‪ .‬اﯾن اﻣر ﻣﯽﺗواﻧد ﺑر‬‫اﺛر ﻓﺷﺎر ﭘﯾوﺳﺗﮫ ﺑﺎزوﺑﻧد ﺑﺎﻋث اﯾﺟﺎد آﺳﯾب ﺷود‪.‬‬‫دﺳﺗﮕﺎه و ﺑﺎزوﺑﻧد در ﺑراﺑر آب ﻣﻘﺎوم ﻧﯾﺳﺗﻧد‪ .‬ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ آب‪ ،‬ﺑﺎران و ﻋرق روی دﺳﺗﮕﺎه و ﺑﺎزوﺑﻧد‬‫ﻟﮑﮫ اﯾﺟﺎد ﻧﮑﻧﻧد‪.‬‬‫اﮔر از دﺳﺗﮕﺎه در ﻧزدﯾﮑﯽ ﺗﻠوﯾزﯾون‪ ،‬ﻓر ﻣﺎﯾﮑرووﯾو‪ ،‬ﺗﻠﻔن ھﻣراه‪ ،‬اﺷﻌﮫ اﯾﮑس ﯾﺎ در ﻧزدﯾﮑﯽ دﯾﮕر‬‫اﺑزارھﺎﯾﯽ اﺳﺗﻔﺎده ﮐﻧﯾد ﮐﮫ ﻣﯾدان اﻟﮑﺗرﯾﮑﯽ ﻗوی دارﻧد‪ ،‬ﻣﻣﮑن اﺳت در اﻧدازهﮔﯾری اﺧﺗﻼل اﯾﺟﺎد ﺷود‪.‬‬‫اﺑزارھﺎی ارﺗﺑﺎطﯽ ﺑﯽﺳﯾم ﻣﺎﻧﻧد دﺳﺗﮕﺎهھﺎی ﺷﺑﮑﮫ ﺧﺎﻧﮕﯽ و ﺗﻠﻔنھﺎی ھﻣراه‪ ،‬ﺗﻠﻔنھﺎی ﺑﯽﺳﯾم و اﯾﺳﺗﮕﺎهھﺎی‬‫ﭘﺎﯾﮫ آﻧﮭﺎ‪ ،‬و ﺗﻠﻔنھﺎی ﺳﯾﺎر ﻣﯽﺗواﻧﻧد ﺑر اﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺗﺄﺛﯾر ﺑﮕذارﻧد‪.‬‬‫ﻟذا ﻻزم اﺳت ﮐﮫ ﺣداﻗل ‪ 30‬ﺳﺎﻧﺗﯽﻣﺗر از اﯾن اﺑزارھﺎ ﻓﺎﺻﻠﮫ داﺷﺗﮫ ﺑﺎﺷﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪2‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد ﻣﺟددا ً از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬از ﺗﻣﯾز ﺑودن دﺳﺗﮕﺎه ﻣطﻣﺋن ﺷوﯾد‪.‬‬‫ﺗﺟﮭﯾزات‪ ،‬ﻗطﻌﺎت و ﺑﺎطریھﺎﯾﯽ ﮐﮫ اﺳﺗﻔﺎده ﺷدهاﻧد‪ ،‬ﻧﺑﺎﯾد ﺑﮫﻋﻧوان ﺿﺎﯾﻌﺎت ﻋﺎدی ﺧﺎﻧﮕﯽ ﻗﻠﻣداد ﺷوﻧد‪،‬‬‫و ﺑﺎﯾد ﺑر اﺳﺎس ﻣﻘررات ﻣﺣﻠﯽ دور اﻧداﺧﺗﮫ ﺷوﻧد‪.‬‬‫وﻗﺗﯽ از آداﭘﺗور ‪ AC‬اﺳﺗﻔﺎده ﻣﯽﮐﻧﯾد‪ ،‬ﻣطﻣﺋن ﺷوﯾد ﮐﮫ آداﭘﺗور ‪ AC‬ﻣﯽﺗواﻧد در ﺻورت ﻟزوم ﺑﮫ‬‫ﺳﮭوﻟت از ﭘرﯾز ﺑرق ﺧﺎرج ﺷود‪.‬‬‫ﺗﻐﯾﯾری در دﺳﺗﮕﺎه اﯾﺟﺎد ﻧﮑﻧﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺑﮫ دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﺑﮫﻣﻧظور اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪ ،‬ﺑﺎزو ﺑﺎﯾد ﺗوﺳط ﺑﺎزوﺑﻧد ﺑﮫﻗدری ﻓﺷرده ﺷود ﮐﮫ ﻣوﻗﺗﺎ ً ﺟرﯾﺎن ﺧون از‬‫ﻣﯾﺎن ﺳرﺧرگ را ﻣﺗوﻗف ﮐﻧد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد درد‪ ،‬ﮐرﺧﺗﯽ ﯾﺎ ﻗرﻣزی ﻣوﻗت ﺑﺎزو ﺷود‪.‬‬‫ﭼﻧﯾن ﺷراﯾطﯽ ﺑﮫوﯾژه زﻣﺎﻧﯽ اﯾﺟﺎد ﻣﯽﺷود ﮐﮫ اﻧدازهﮔﯾری را ﭼﻧد ﻣرﺗﺑﮫ ﭘﺷت ﺳرھم اﻧﺟﺎم ﻣﯽدھﯾد‪.‬‬‫ھرﮔوﻧﮫ درد‪ ،‬ﮐرﺧﺗﯽ ﯾﺎ ﻗرﻣزی ﺑﺎ ﮔذﺷت زﻣﺎن از ﺑﯾن ﻣﯽرود‪.‬‬‫اﮔر اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﭼﻧدﯾن ﻣرﺗﺑﮫ ﺗﮑرار ﺷود‪ ،‬ﻣﯽﺗواﻧد ﺑﮫ دﻟﯾل ﺗداﺧﻠﯽ ﮐﮫ در ﺟرﯾﺎن ﺧون اﯾﺟﺎد‬‫ﻣﯽﮐﻧد‪ ،‬ﺑﮫ ﻓرد آﺳﯾب ﺑزﻧد‪ .‬اﯾن ﻣوﺿوع را ﺑررﺳﯽ ﮐﻧﯾد ﮐﮫ در ﺻورت ﺗﮑرار اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪،‬‬‫ﻋﻣﻠﯾﺎت دﺳﺗﮕﺎه ﻣوﺟب اﺧﺗﻼل طوﻻﻧﯽ ﻣدت در ﺟرﯾﺎن ﺧون ﻧﺷود‪.‬‬‫اﮔر ﻋﻣل ﻣﺎﺳﺗﮑﺗوﻣﯽ داﺷﺗﮫاﯾد و ﺳﯾﻧﮫﺗﺎن ﺗﺧﻠﯾﮫ ﺷده اﺳت‪ ،‬ﭘﯾش از اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪.‬‬‫اﺟﺎزه ﻧدھﯾد ﮐﮫ ﮐودﮐﺎن از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﻧد و در ﺟﺎﯾﯽ ﮐﮫ در دﺳﺗرس اطﻔﺎل اﺳت از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده‬‫ﻧﮑﻧﯾد‪ .‬ﻋدم ﺗوﺟﮫ ﺑﮫ اﯾن ﻣوﺿوع ﻣﯽﺗواﻧد ﺑﺎﻋث ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺷود‪.‬‬‫اﺟزای ﮐوﭼﮑﯽ در اﯾن دﺳﺗﮕﺎه وﺟود دارد ﮐﮫ در ﺻورت ﺑﻠﻌﯾده ﺷدن ﺗوﺳط اطﻔﺎل ﻣﯽﺗواﻧد ﺑﺎﻋث ﺧﻔﮕﯽ‬‫آﻧﮭﺎ ﺷود‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ دﯾﮕر ﻧﻣﯽﺧواھﯾد از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬آداﭘﺗور ‪ AC‬را ﺧﺎرج ﮐﻧﯾد‪.‬‬‫اﺳﺗﻔﺎده از ﻟوازم ﺟﺎﻧﺑﯽ ﮐﮫ در اﯾن راھﻧﻣﺎ ﺑﮫ آﻧﮭﺎ اﺷﺎره ﻧﺷده اﺳت ﻣﯽﺗواﻧد ﺑﺎﻋث آﺳﯾب ﺑﮫ ﻓرد و دﺳﺗﮕﺎه‬‫ﺷود‪.‬‬‫از آﻧﺟﺎﯾﯽ ﮐﮫ ﺑﺎطری دارای ﻣدار ﮐوﺗﺎه اﺳت‪ ،‬ﻣﯽﺗواﻧد داغ ﺷده و ﺑﺎﻋث اﯾﺟﺎد ﺳوﺧﺗﮕﯽ ﺷود‪.‬‬‫ﭘﯾش از ﺷروع ﮐﺎر‪ ،‬اﺟﺎزه دھﯾد ﮐﮫ دﺳﺗﮕﺎه ﺧودش را ﺑﺎ ﻣﺣﯾط اطراف ﺳﺎزﮔﺎر ﮐﻧد )ﺣدود ﯾﮏ ﺳﺎﻋت(‪.‬‬‫آزﻣﺎﯾش ﺑﺎﻟﯾﻧﯽ روی ﻧوزادان و زﻧﺎن ﺑﺎردار اﻧﺟﺎم ﻧﺷده اﺳت‪ .‬ﻟذا از اﯾن دﺳﺗﮕﺎه ﺑرای ﻧوزادان و زﻧﺎن‬‫ﺑﺎردار اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪.‬‬‫ﺑﺎطریھﺎ‪ ،‬ﻓﯾش ‪ DC‬و ﺧود ﺑﯾﻣﺎر را ھﻣزﻣﺎن ﻟﻣس ﻧﮑﻧﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺑرقﮔرﻓﺗﮕﯽ ﺷود‪.‬‬‫از ﺑﺎد ﮐردن دﺳﺗﮕﺎه ﺑدون ﭘﯾﭼﺎﻧدن آن ﺑﮫ دور ﺑﺎزو اﺟﺗﻧﺎب ﮐﻧﯾد‪.‬‬‫ﻣوارد ﻣﻧﻊ اﺳﺗﻔﺎده‬‫در اﯾن ﺑﺧش ﻣوارد اﺣﺗﯾﺎطﯽ ﺑرای اﺳﺗﻔﺎده ﺻﺣﯾﺢ از دﺳﺗﮕﺎه ذﮐر ﺷده اﺳت‪.‬‬‫ﺑﺎزوﺑﻧد دﺳﺗﮕﺎه ﻓﺷﺎر ﺧون را روی ﺑﺎزوﯾﯽ ﮐﮫ ﺗﺟﮭﯾزات اﻟﮑﺗرﯾﮑﯽ ﭘزﺷﮑﯽ دﯾﮕری روی آن ﻣﺗﺻل‬‫اﺳت‪ ،‬ﺑﮫ ﮐﺎر ﻧﺑرﯾد‪ .‬در ﭼﻧﯾن ﺷراﯾطﯽ ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫﺧوﺑﯽ ﮐﺎر ﻧﮑﻧد‪.‬‬‫اﻓرادی ﮐﮫ دﭼﺎر ﻧﻘص ﺷدﯾد ﮔردش ﺧون در ﺑﺎزوھﺎﯾﺷﺎن ھﺳﺗﻧد ﺑﺎﯾد ﭘﯾش از اﺳﺗﻔﺎده از اﯾن دﺳﺗﮕﺎه ﺑﺎ‬‫ﭘزﺷﮑﺷﺎن ﻣﺷورت ﮐﻧﻧد ﺗﺎ از اﯾﺟﺎد ﻣﺷﮑﻼت ﭘزﺷﮑﯽ اﺟﺗﻧﺎب ﺷود‪.‬‬‫از ﺗﺷﺧﯾص ﺧودﺳراﻧﮫ ﺑر اﺳﺎس ﻧﺗﺎﯾﺞ اﻧدازهﮔﯾری ﺑﭘرھﯾزﯾد و از ھرﮔوﻧﮫ ﺧوددرﻣﺎﻧﯽ اﺟﺗﻧﺎب ﮐﻧﯾد‪.‬‬‫ھﻣﯾﺷﮫ ﺑرای ارزﯾﺎﺑﯽ ﻧﺗﯾﺟﮫ و ﺗﺟوﯾز درﻣﺎن ﻣﻧﺎﺳب ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪.‬‬‫ﺑﺎزوﺑﻧد دﺳﺗﮕﺎه ﻓﺷﺎر ﺧون را روی ﺑﺎزوﯾﯽ ﮐﮫ زﺧم درﻣﺎن ﻧﺷده دارد ﻗرار ﻧدھﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪3‬‬‫ﺑﺎزوﺑﻧد دﺳﺗﮕﺎه ﻓﺷﺎر ﺧون را روی ﺑﺎزوﯾﯽ ﮐﮫ ﺗﺣت ﺗزرﯾﻖ ورﯾدی ﯾﺎ اﻧﺗﻘﺎل ﺧون اﺳت‪ ،‬ﻗرار ﻧدھﯾد‪ .‬اﯾن‬‫ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺷود‪.‬‬‫در ﺟﺎﯾﯽ ﮐﮫ ﮔﺎزھﺎی اﺷﺗﻌﺎلﭘذﯾر ﻣﺎﻧﻧد ﮔﺎزھﺎی ﺑﯽھوﺷﯽ وﺟود دارد ﻧﺑﺎﯾد از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﺷود‪.‬‬‫اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﻧﻔﺟﺎر ﺷود‪.‬‬‫در ﻣﺣﯾطھﺎﯾﯽ ﻣﺎﻧﻧد اﺗﺎﻗﮏ اﮐﺳﯾژن ﻓﺷﺎر ﺑﺎﻻ ﯾﺎ ﭼﺎدر اﮐﺳﯾژن ﮐﮫ اﮐﺳﯾژن ﺑﺳﯾﺎر ﻣﺗراﮐم دارﻧد‪ ،‬ﻧﺑﺎﯾد از‬‫اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﺷود‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث آﺗشﺳوزی ﯾﺎ اﻧﻔﺟﺎر ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪4‬‬‫ﻣﻌرﻓﯽ ﻗطﻌﺎت‬‫ﻧﻣﺎﯾﺷﮕر‬‫ﻧری راﺑط ھوا‬‫ﻧری آداﭘﺗور‪AC‬‬‫ﻓﯾش ‪DC‬‬‫دﮐﻣﮫ ‪START‬‬‫ﻣﺎدﮔﯽ ھوا‬‫ﺷﯾﻠﻧﮓ ھوا‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﻧﺷﺎﻧﮫ ﺷﺎﺧص‬‫ﺑﺎزوﺑﻧد‬‫ﻣﺣﻔظﮫ ﺑﺎطری‬‫درﭘوش ﺑﺎطری‬‫ﺑﺎطریھﺎی ‪ 1.5‬وﻟﺗﯽ‬‫)‪ R6P،LR6‬ﯾﺎ ‪(AA‬‬‫ﻧﻣﺎﯾﺷﮕر‬‫ﻣﯾﺎﻧﮕﯾن‬‫ﺣﺎﻓظﮫ‬‫ﻧﻣﺎد ‪IHB/AFib‬‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫‪%IHB/AFib‬‬‫ﻧﻣﺎد ﺧطﺎی اﻧدازه ﺑﺎزوﺑﻧد‬‫ﻧﻣﺎد ﺧطﺎی ﻧﺎﺷﯽ از ﺣرﮐت‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫ﻋﻼﻣت ﻗﻠب‬‫ﺿرﺑﺎن ﻗﻠب‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫و ﺷﺎﺧص ﺑﺎر ﻓﺷﺎر‬‫ﺷﺎﺧص ﺑﺎطری‬‫ﻓﺎرﺳﯽ ‪5‬‬‫ﻧﻣﺎدھﺎ‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی ﺟﻌﺑﮫ دﺳﺗﮕﺎه ﭼﺎپ ﺷدهاﻧد‬‫ﻧﻣﺎدھﺎ‬‫‪SYS‬‬‫‪DIA‬‬‫‪PUL‬‬‫‪IP‬‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﺣﺎﻟت اﻧﺗظﺎر و روﺷن ﮐردن دﺳﺗﮕﺎه‪.‬‬‫ﻓﺷﺎر ﺧون ﺳﯾﺳﺗوﻟﯽ ﺑر ﺣﺳب ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﻓﺷﺎر ﺧون دﯾﺎﺳﺗوﻟﯽ ﺑر ﺣﺳب ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﺿرﺑﺎن ﻗﻠب در ھر دﻗﯾﻘﮫ‬‫راھﻧﻣﺎی ﻧﺻب ﺑﺎطری‬‫ﺟرﯾﺎن ﻣﺳﺗﻘﯾم‬‫ﻧوع ‪ :BF‬اﺑزار‪ ،‬ﺑﺎزوﺑﻧد و ﻟوﻟﮫھﺎ ﺑﮫﮔوﻧﮫای طراﺣﯽ ﺷدهاﻧد ﮐﮫ از ﺑروز ﺑرقﮔرﻓﺗﮕﯽ‬‫ﺟﻠوﮔﯾری ﺷود‪.‬‬‫ﺑرﭼﺳب دﺳﺗور ‪ EC‬اﺑزار ﭘزﺷﮑﯽ‬‫ﻧﻣﺎﯾﻧده ‪EU‬‬‫ﺗوﻟﯾدﮐﻧﻧده‬‫ﺗﺎرﯾﺦ ﺗوﻟﯾد‬‫ﻧﻣﺎد ﺣﻔﺎظت ﺑﯾناﻟﻣﻠﻠﯽ‬‫ﺑرﭼﺳب ‪WEEE‬‬‫‪SN‬‬‫ﺷﻣﺎره ﺳ﷼‬‫ﺑﮫ ﮐﺗﺎﺑﭼﮫ‪/‬دﻓﺗرﭼﮫ راھﻧﻣﺎ ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫ﻗطﺑﯾت ﻓﯾش ‪DC‬‬‫در ﺟﺎی ﺧﺷﮏ ﻧﮕﺎه دارﯾد‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی ﻧﻣﺎﯾﺷﮕر ظﺎھر ﻣﯽﺷوﻧد‬‫ﻧﻣﺎدھﺎ‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫ ﺷده‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‪ .‬ﺗﺎ‬‫زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم‬‫ﺟﺎﯾﯽ ﮐﮫ ﻣﯽﺗواﻧﯾد ﺑﯽﺣرﮐت‬‫اﺳت‪ .‬ھﻧﮕﺎﻣﯽ ﮐﮫ ﻧﺑض ﺷﻧﺎﺳﺎﯾﯽ ﺷد‪ ،‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﻧﻣﺎد ‪IHB/AFib‬‬‫زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ ﺿرﺑﺎن ﻗﻠب ﻧﺎﻣﻧظم ﺷﻧﺎﺳﺎﯾﯽ‬‫ﺷود‪.‬‬‫اﯾن ﻧﻣﺎد زﻣﺎﻧﯽ روﺷن ﻣﯽﺷود ﮐﮫ ﺗﮑﺎنھﺎی ﺑﺳﯾﺎر‬‫آراﻣﯽ ﻣﺎﻧﻧد ﻟرزش ﺷﻧﺎﺳﺎﯾﯽ ﺷود‪.‬‬‫ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﻋدد ﻧﺎدرﺳﺗﯽ‬‫را ﻧﺷﺎن دھد‪ .‬اﻧدازهﮔﯾری را‬‫زﻣﺎﻧﯽ ظﺎھرا ً ﻣﯽﺷود ﮐﮫ ﺣرﮐﺗﯽ در ﺑدن ﯾﺎ ﺑﺎزو‬‫ﺗﮑرار ﮐﻧﯾد‪.‬‬‫ﺷﻧﺎﺳﺎﯾﯽ ﺷود‪.‬‬‫در طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪6‬‬‫ﻧﻣﺎدھﺎ‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫زﻣﺎﻧﯽ آﺷﮑﺎر ﻣﯽﺷود ﮐﮫ در ﺣﯾن اﻧدازهﮔﯾری‬‫ﺑﺎزوﺑﻧد ﺷل ﺑﺎﺷد‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫ ﺷده‬‫ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﻋدد ﻧﺎدرﺳﺗﯽ را‬‫ﻧﺷﺎن دھد‪ .‬ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ ﺑﺑﻧدﯾد‪،‬‬‫و اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫ﺳرﻋت ﺷﻧﺎﺳﺎﯾﯽﺷده ‪ IHB/AFib‬در ﺣﺎﻓظﮫ‬‫]‪x100 [%‬‬

‫ﺗﻌداد ‪ IHB/AFibs‬ﺷﻧﺎﺳﺎﯾﯽﺷده‬‫در ﺣﺎﻓظﮫ‬‫ﺗﻌداد ﮐل‬‫= ‪%IHB / AFib‬‬‫اﻧدازهﮔﯾریھﺎی ﻗﺑﻠﯽ ﮐﮫ در ﺣﺎﻓظﮫ ذﺧﯾره ﺷده‬‫اﺳت‪.‬‬‫ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت‬‫‪) FULL BATTERY‬ﺑﺎطری ﭘر(‬‫ﻧﺷﺎﻧﮕر ﺗوان ﺑﺎطری در طول اﻧدازهﮔﯾری‪.‬‬‫‪) LOW BATTERY‬ﺑﺎطری ﮐم(‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﺎطری ﮐم ﺷود‪ ،‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ اﯾن ﻧﺷﺎﻧﮫ ﭼﺷﻣﮏ ﻣﯽزﻧد‪،‬‬‫ھﻣﮫ ﺑﺎطریھﺎ را ﺗﻌوﯾض ﮐﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫ﻓﺷﺎر ﺧون ﻧﺎﭘﺎﯾدار ﺑﮫ دﻟﯾل ﺗﮑﺎن ﺧوردن در طول‬‫در طول اﻧدازهﮔﯾری ﮐﺎﻣﻼً ﺑﯽﺣرﮐت‬‫اﻧدازهﮔﯾری‪.‬‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﻣﻘﺎدﯾر ﺳﯾﺳﺗوﻟﯽ و دﯾﺎﺳﺗوﻟﯽ درون ‪ ۱۰‬ﻣﯾﻠﯽﻣﺗر‬‫ﺟﯾوه از ھﻣدﯾﮕر ھﺳﺗﻧد‪.‬‬‫ﻣﻘدار ﻓﺷﺎر در طول ﺑﺎد ﮐردن اﻓزاﯾش ﻧﯾﺎﻓت‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ ﺑﺑﻧدﯾد‪ ،‬و‬‫ﺑﺎزوﺑﻧد ﺑﮫدرﺳﺗﯽ ﻗرار ﻧﮕرﻓﺗﮫ اﺳت‪.‬‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫‪PUL DISPLAY ERROR‬‬‫)ﺧطﺎی ﻧﻣﺎﯾش(‬‫ﭘﺎﻟس ﺑﮫدرﺳﺗﯽ ﺷﻧﺎﺳﺎﯾﯽ ﻧﺷده اﺳت‪.‬‬‫ﺑﺎطریھﺎ را ﺧﺎرج ﮐرده و دﮐﻣﮫ‬‫‪ START‬را ﺑزﻧﯾد‪ ،‬و ﺑﺎر دﯾﮕر‬‫ﺑﺎطریھﺎ را ﻧﺻب ﮐﻧﯾد‪ .‬اﮔر ﻧﺷﺎﻧﮫ‬‫ﺧطﺎی داﺧﻠﯽ دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون‬‫ﺧطﺎ ھﻣﭼﻧﺎن ظﺎھر ﺷد‪ ،‬ﺑﺎ ﻓروﺷﻧده‬‫ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪7‬‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت‬‫‪ .1‬اﻧدازهﮔﯾری ﻧرﻣﺎل‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪ .‬ﻓﺷﺎر ﺧون اﻧدازهﮔﯾری ﺷده و اطﻼﻋﺎت در ﺣﺎﻓظﮫ ذﺧﯾره ﻣﯽﺷود‪ .‬اﯾن دﺳﺗﮕﺎه‬‫ﻣﯽﺗواﻧد ‪ ۶۰‬اﻧدازهﮔﯾری آﺧر را در ﺣﺎﻓظﮫاش ذﺧﯾره ﮐﻧد‪.‬‬‫‪ .2‬ﻓراﺧواﻧﯽ اطﻼﻋﺎت‬‫اﮔر ھﯾﭻ ﭼﯾز ﻧﻣﺎﯾش داده ﻧﺷد‪ START ،‬را ﻓﺷﺎر داده و ﻧﮕﺎه‬‫دارﯾد‪.‬‬‫در ﺣﺎﻟت اﻧﺗظﺎر‪،‬‬‫دﮐﻣﮫ را‬‫ﻓﺷﺎر دھﯾد و‬‫ﻧﮕﺎه دارﯾد‬‫‪START‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت ﻧﺷﺎن داده ﺷد‪ ،‬دﮐﻣﮫ را رھﺎ ﮐﻧﯾد‪.‬‬‫ﺷﻣﺎره اطﻼﻋﺎت و اطﻼﻋﺎت ذﺧﯾره ﺷده ﺑﮫﺻورت ﺧودﮐﺎر و‬‫ﺑﮫ ﺗرﺗﯾب از آﺧرﯾن اﻧدازهﮔﯾری ﻧﺷﺎن داده ﻣﯽﺷوﻧد‪.‬‬‫ﺑرای ﻣﺷﺎھده ﺟزﺋﯾﺎت ﺑﯾﺷﺗر در ﻣورد ﻓراﺧواﻧﯽ اطﻼﻋﺎت‪ ،‬ﺑﮫ‬‫"ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ" ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫‪ ۳.‬ﺣذف ﮐﻠﯾﮫ اطﻼﻋﺎت ذﺧﯾرهﺷده در ﺣﺎﻓظﮫ‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ دﺳﺗﮕﺎه را ﺧﺎﻣوش ﮐردﯾد‪ ،‬دﮐﻣﮫ‬‫" ظﺎھر ﺷود‪.‬‬‫‪ START‬را ﻓﺷﺎر داده و ﻧﮕﺎه دارﯾد ﺗﺎ زﻣﺎﻧﯽ ﮐﮫ "‬‫" را اﻧﺗﺧﺎب ﮐﻧﯾد ﺗﺎ اطﻼﻋﺎت ﭘﺎک ﺷود‪.‬‬‫"‬‫ﺑﮫ ﻣﻌﻧﺎی ﭘﺎک ﺷدن اطﻼﻋﺎت اﺳت‪.‬‬‫ﭼﺷﻣﮏ زدن ﻧﺷﺎﻧﮫ‬‫دﺳﺗﮕﺎه ﺑﮫﺻورت ﺧودﮐﺎر ﺧﺎﻣوش ﻣﯽﺷود‪.‬‬‫‪ .۴‬اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫‪START‬‬‫ﺑﮫ ﺻﻔﺣﮫ ‪ ۱۵‬ﻣراﺟﻌﮫ ﮐﻧﯾد ﺗﺎ اطﻼﻋﺎت ﺑﯾﺷﺗری در ﻣورد اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب ﺑﯾﺎﺑﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪8‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﺻب ‪ /‬ﺗﻌوﯾض ﺑﺎطریھﺎ‬‫‪ .1‬درﭘوش ﺑﺎطری را ﺑردارﯾد‪.‬‬‫ﻣرﺣﻠﮫ ‪1‬‬‫‪ .2‬ﺑﺎطریھﺎی ﮐﮭﻧﮫ را ﺑردارﯾد و ھﻣﺎنطور ﮐﮫ ﻧﺷﺎن داده‬‫ﺷده اﺳت‪ ،‬ﺑﺎطریھﺎی ﻧو را در ﻣﺣﻔظﮫ ﺑﺎطری ﻗرار‬‫دھﯾد‪ .‬ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ ﻗطبھﺎی ﺑﺎطری‬‫)‪ +‬و ‪ (-‬در ﺟﺎی ﺻﺣﯾﺢ ﻗرار ﮔﯾرﻧد‪.‬‬‫ﺻرﻓﺎ ً از ﺑﺎطریھﺎی ‪ R6P، LR6‬ﯾﺎ ‪ AA‬اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫ﻣرﺣﻠﮫ ‪3‬‬‫ﻣرﺣﻠﮫ ‪1‬‬‫ﻣرﺣﻠﮫ ‪2‬‬‫‪ .3‬درﭘوش ﺑﺎطری را ﺑﮕذارﯾد‪.‬‬‫اﺣﺗﯾﺎط‬‫ﺑﺎطریھﺎ را ھﻣﺎنﮔوﻧﮫ ﮐﮫ در ﺗﺻوﯾر ﻧﺷﺎن داده ﺷده اﺳت در ﻣﺣﻔظﮫ ﺑﺎطری ﻗرار دھﯾد‪ .‬اﮔر ﺑﺎطریھﺎ ﺑﮫ‬‫ﺷﮑل ﻧﺎدرﺳﺗﯽ ﻗرار ﮔﯾرﻧد‪ ،‬دﺳﺗﮕﺎه ﮐﺎر ﻧﺧواھد ﮐرد‪.‬‬‫)‪ (LOW BATTERY‬ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐرد‪ ،‬ھﻣﮫ ﺑﺎطریھﺎ را ﺑﺎ‬‫وﻗﺗﯽ ﻋﻼﻣت‬‫ﺑﺎطریھﺎی ﻧو ﺗﻌوﯾض ﮐﻧﯾد‪ .‬ﺑﺎطریھﺎی ﮐﮭﻧﮫ و ﻧو را ﺑﺎ ھم در دﺳﺗﮕﺎه ﻗرار ﻧدھﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد‬‫ﻣوﺟب ﮐوﺗﺎه ﺷدن ﻋﻣر ﺑﺎطری ﯾﺎ ﻧﻘص در ﮐﺎرﮐرد دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﺑﺎطریھﺎ را دو ﯾﺎ ﭼﻧد دﻗﯾﻘﮫ ﺑﻌد از ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه ﺗﻌوﯾض ﮐﻧﯾد‪.‬‬‫)‪ (LOW BATTERY‬ﺣﺗﯽ ﭘس از ﺗﻌوﯾض ﺑﺎطریھﺎ ظﺎھر ﺷد‪ ،‬ﯾﮏﺑﺎر اﻧدازهﮔﯾری‬‫اﮔر ﻋﻼﻣت‬‫ﻓﺷﺎر ﺧون را اﻧﺟﺎم دھﯾد‪ .‬آﻧﮕﺎه دﺳﺗﮕﺎه ﻣﯽﺗواﻧد ﺑﺎطریھﺎی ﺟدﯾد را ﺷﻧﺎﺳﺎﯾﯽ ﮐﻧد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﺎطری ﺧﺎﻟﯽ ﮐرده ﺑﺎﺷد‪ ،‬ﻋﻼﻣت )‪ (LOW BATTERY‬ظﺎھر ﻧﻣﯽﺷود‪.‬‬‫ﻋﻣر ﺑﺎطری ﺑﺳﺗﮫ ﺑﮫ دﻣﺎی ﭘﯾراﻣوﻧﯽ ﻣﯽﺗواﻧد ﻣﺗﻐﯾر ﺑﺎﺷد و در دﻣﺎی ﮐم ﻣﻣﮑن اﺳت ﻋﻣر ﺑﺎطری ﮐﻣﺗر‬‫ﺷود‪ .‬ﺑﮫطور ﮐﻠﯽ‪ ،‬ﭼﮭﺎر ﺑﺎطری ﻧوی ‪ LR6‬در دﺳﺗﮕﺎھﯽ ﮐﮫ دو ﺑﺎر در روز اﺳﺗﻔﺎده ﻣﯽﺷود‪ ،‬ﺣدودا ً ﯾﮏ‬‫ﺳﺎل ﻋﻣر ﻣﯽﮐﻧد‪.‬‬‫ﺻرﻓﺎ ً از ﺑﺎطریھﺎی ﺗﻌﯾﯾنﺷده اﺳﺗﻔﺎده ﮐﻧﯾد‪ .‬ﺑﺎطریھﺎﯾﯽ ﮐﮫ ھﻣراه ﺑﺎ دﺳﺗﮕﺎه ﺑﮫ ﺷﻣﺎ داده ﻣﯽﺷوﻧد‪ ،‬ﺑرای‬‫آزﻣﺎﯾش ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ھﺳﺗﻧد و ﻟذا ﻣﻣﮑن اﺳت ﻋﻣر ﻣﺣدودی داﺷﺗﮫ ﺑﺎﺷﻧد‪.‬‬‫اﮔر ﻗرار اﺳت ﮐﮫ از دﺳﺗﮕﺎه ﺑرای ﻣدت طوﻻﻧﯽ اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪ ،‬ﺑﺎطریھﺎﯾش را ﺧﺎرج ﮐﻧﯾد‪.‬‬‫ﻣﻣﮑن اﺳت ﺑﺎطری ﻧﺷت ﮐﻧد و ﺑﺎﻋث اﺧﺗﻼل در ﮐﺎرﮐرد دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪9‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻣﺗﺻل ﮐردن ﺷﯾﻠﻧﮓ ھوا‬‫ﻣﺎدﮔﯽ ھوا‬‫ﻣﺎدﮔﯽ ‪r‬‬‫ﻧری راﺑط ھوا را ﻣﺣﮑم ﺑﮫ داﺧل ﻣﺎدﮔﯽ ھوا‬‫وارد ﮐﻧﯾد‪.‬‬‫ﻧری راﺑط ھوا‬‫ﻣﺗﺻل ﮐردن آداﭘﺗور ‪AC‬‬‫ﻧری آداﭘﺗور ‪ AC‬را ﺑﮫ ﻓﯾش ‪ DC‬وارد ﮐﻧﯾد‪.‬‬‫ﺳﭘس‪ ،‬آداﭘﺗور ‪ AC‬را ﺑﮫ ﭘرﯾز ﺑرق ﻣﺗﺻل ﮐﻧﯾد‪.‬‬‫از آداﭘﺗور ‪ AC‬ﻣﺷﺧصﺷده اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫)ﺑﮫ ﺻﻔﺣﮫ ‪ 22‬ﻣراﺟﻌﮫ ﮐﻧﯾد(‬‫ﻓﯾش ‪DC‬‬‫ﻧری آداﭘﺗور ‪AC‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد آداﭘﺗور ‪ AC‬را از ﭘرﯾز ﺑرق ﺟدا ﮐﻧﯾد‪ ،‬ﺑدﻧﮫ آداﭘﺗور ‪ AC‬را ﺑﮕﯾرﯾد و ﺑﮑﺷﯾد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد ﻧری آداﭘﺗور ‪ AC‬را از دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺟدا ﮐﻧﯾد‪ ،‬ﻧری آداﭘﺗور ‪AC‬‬‫را ﮔرﻓﺗﮫ و از دﺳﺗﮕﺎه ﺑﯾرون ﺑﮑﺷﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪10‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫اﻧﺗﺧﺎب ﺳﺎﯾز ﻣﻧﺎﺳب ﺑﺎزوﺑﻧد‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧدی ﺑﺎ اﻧدازه ﻣﻧﺎﺳب ﻧﻘش ﻣﮭﻣﯽ در ﻗراﺋت ﺻﺣﯾﺢ دﺳﺗﮕﺎه دارد‪ .‬اﮔر ﺑﺎزوﺑﻧد اﻧدازه ﻣﻧﺎﺳب‬‫ﻧداﺷﺗﮫ ﺑﺎﺷد‪ ،‬ﻣﻣﮑن اﺳت ﻗراﺋت دﺳﺗﮕﺎه اﺷﺗﺑﺎه ﺷود‪.‬‬‫اﻧدازه ﺑﺎزو روی ھر ﺑﺎزوﺑﻧد ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺷﺎﺧص و داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب روی ﺑﺎزوﺑﻧد ﺑﮫ ﺷﻣﺎ ﻣﯽﮔوﯾد ﮐﮫ آﯾﺎ از ﺑﺎزوﺑﻧد ﺻﺣﯾﺣﯽ اﺳﺗﻔﺎده ﻣﯽﮐﻧﯾد ﯾﺎ‬‫ﺧﯾر‪) .‬ﺑﮫ ﺑﺧش "ﻧﻣﺎدھﺎی ﭼﺎپﺷده روی ﺑﺎزوﺑﻧد" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد(‬‫اﮔر ﺷﺎﺧص ﺧﺎرج از داﻣﻧﮫ را ﻧﺷﺎن داد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽﺗﺎن ﺗﻣﺎس ﺑﮕﯾرﯾد ﺗﺎ ﺑﺎزوﺑﻧد دﯾﮕری ﺑﮫ ﺷﻣﺎ‬‫ﺑدھد‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫﻋﻧوان ﯾﮏ ﮐﺎﻻی ﻣﺻرﻓﯽ ﻗﻠﻣداد ﻣﯽﺷود‪ .‬اﮔر ﺑﺎزوﺑﻧد ﭘﺎره ﺷد‪ ،‬ﯾﮑﯽ دﯾﮕر ﺑﺧرﯾد‪.‬‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫اﻧدازه ﺗوﺻﯾﮫ ﺷده ﺑﺎزوﺑﻧد‬‫اﻧدازه ﺑﺎزو‬‫ﺑﺎزوﺑﻧد ﺑزرگ ﺑزرﮔﺳﺎﻻن‬‫‪ 31‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 45‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 42‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 32‬ﺳﺎﻧﺗﯽﻣﺗر‬‫اﻧدازه ﺑﺎزو‪ :‬ﻣﺣﯾط ﺑﺎزو در ﺑﺧش ﻋﺿﻠﮫ دو ﺳر‪.‬‬‫‪ UA-767S-W‬ﺑرای اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد ﮐوﭼﮏ طراﺣﯽ ﻧﺷده اﺳت‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد‬‫‪ .1‬ھﻣﺎنطور ﮐﮫ در ﺗﺻوﯾر ﻣﺷﺎھده ﻣﯽﮐﻧﯾد‪ ،‬ﺑﺎزوﺑﻧد‬‫را دور ﺑﺎزو‪ ،‬ﯾﻌﻧﯽ ‪ 1‬ﺗﺎ ‪ 2‬ﺳﺎﻧﺗﯽﻣﺗر ﺑﺎﻻﺗر از‬‫ﮔودی آرﻧﺞ ﺑﺑﻧدﯾد‪.‬‬‫ً‬‫ﺑﺎزوﺑﻧد را ﻣﺳﺗﻘﯾﻣﺎ روی ﭘوﺳت ﻗرار دھﯾد‪ ،‬زﯾرا‬‫ﻟﺑﺎس ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺿرﺑﺎن ﺿﻌﯾف ﺷود و‬‫در ﻗراﺋت ﺧطﺎ اﯾﺟﺎد ﮐﻧد‪.‬‬‫‪ .2‬ﺗﻧﮓ ﺷدن ﺑﺎزو ﺑر اﺛر ﺑﺎﻻ زدن زﯾﺎد آﺳﺗﯾن ﻣﯽﺗواﻧد‬‫ﺑﺎﻋث اﯾﺟﺎد ﺧطﺎ در ﻗراﺋت دﺳﺗﮕﺎه ﺷود‪.‬‬‫‪ .3‬ﻣطﻣﺋن ﺷوﯾد ﮐﮫ ﺷﺎﺧص‬‫ﻣﻧﺎﺳﺑﯽ ﻗرار دارد‪.‬‬‫آﺳﺗﯾنھﺎ را زﯾﺎد ﺑﺎﻻ ﻧزﻧﯾد‪.‬‬‫‪ 1 - 2‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد‬‫ﺷﯾﻠﻧﮓ ھوا‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﭼﺳب ﺑﺎزوﺑﻧد را ﻣﺣﮑم ﮐﻧﯾد‪.‬‬‫در داﻣﻧﮫ اﻧدازه‬‫ﺷﺎﺧص‬‫ﻓﺎرﺳﯽ ‪11‬‬‫و داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﻣﺎدھﺎی ﭼﺎپ ﺷده روی ﺑﺎزوﺑﻧد‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫▲‬‫‪REF‬‬‫ﺷﺎﺧص‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‪.‬‬‫روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫ ﺷده‬‫ﻧﺷﺎﻧﮫ را روی ﺷرﯾﺎن ﺑﺎزو ﯾﺎ‬‫ھمراﺳﺗﺎ ﺑﺎ اﻧﮕﺷت ﺣﻠﻘﮫ در داﺧل‬‫ﺑﺎزو ﺗﻧظﯾم ﮐﻧﯾد‪.‬‬‫از ﺑﺎزوﺑﻧد ﺑزرگ ﺑزرﮔﺳﺎﻻن ﺑﮫﺟﺎی‬‫ﺧﺎرج از داﻣﻧﮫ ﮐﮫ روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎل و ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﯾﺎ ﺑﺎزوﺑﻧد ﻋرﯾض‬‫ﻋرﯾض ﭼﺎپ ﻣﯽﺷود‪.‬‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﻋرﯾض‪ .‬روی ﺑﺎزوﺑﻧد‬‫ﻋرﯾض ﭼﺎپ ﺷده اﺳت‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ‬‫ﺑزرﮔﺳﺎﻻن‪ .‬روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎل ﭼﺎپ‬‫ﺷده اﺳت‪.‬‬‫زﯾر داﻣﻧﮫ ﮐﮫ روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن و ﺑﺎزوﺑﻧد‬‫ﻋرﯾض ﭼﺎپ ﺷده اﺳت‪.‬‬‫زﯾر داﻣﻧﮫ ﮐﮫ روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن از ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﺑﮫﺟﺎی ﺑﺎزوﺑﻧد‬‫ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺷﻣﺎره دﺳﺗﮫ‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن‬

‫‪L‬‬

‫‪A‬‬

‫‪L‬‬

‫‪W‬‬

‫‪S‬‬

‫‪L‬‬

‫‪A‬‬

‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫ﻓﺎرﺳﯽ ‪12‬‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪S‬‬

‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﺣوه اﻧﺟﺎم اﻧدازهﮔﯾری دﻗﯾﻖ‬‫ﺑﮫﻣﻧظور اﻧﺟﺎم دﻗﯾﻖﺗرﯾن اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪:‬‬‫آرام روی ﺻﻧدﻟﯽ ﺑﻧﺷﯾﻧﯾد‪ .‬دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﺑﮕذارﯾد‪ .‬ﭘﺎھﺎﯾﺗﺎن را رویھم ﻧﯾﻧدازﯾد‪ .‬ﭘﺎھﺎﯾﺗﺎن را‬‫ﺻﺎف روی زﻣﯾن ﺑﮕذارﯾد و ﮐﻣرﺗﺎن را ﺻﺎف ﮐﻧﯾد‪.‬‬‫ﺑﮫ ﻣدت ﭘﻧﺞ اﻟﯽ ده دﻗﯾﻘﮫ آرام ﺑﺎﺷﯾد و ﺳﭘس اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫ﻣرﮐز ﺑﺎزوﺑﻧد را ھمﺗراز ﺑﺎ ﻗﻠﺑﺗﺎن ﻗرار دھﯾد‪.‬‬‫در طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت و آرام ﺑﻣﺎﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻧﺑﺎﯾد ﺑﻼﻓﺎﺻﻠﮫ ﭘس از ورزش ﯾﺎ ﺣﻣﺎم ﮐردن اﻧﺟﺎم ﺷود‪ .‬ﺑﯾﺳت اﻟﯽ ﺳﯽ دﻗﯾﻘﮫ آرام ﺑﻣﺎﻧﯾد و ﺑﻌد‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫ﺳﻌﯽ ﮐﻧﯾد اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون را ھر روز در ﯾﮏ ﺳﺎﻋت ﻣﻌﯾن اﻧﺟﺎم دھﯾد‪.‬‬‫اﻧدازهﮔﯾری‬‫در طول اﻧدازهﮔﯾری‪ ،‬طﺑﯾﻌﯽ اﺳت ﮐﮫ اﺣﺳﺎس ﮐﻧﯾد ﺑﺎزوﺑﻧد ﺧﯾﻠﯽ ﻣﺣﮑم ﺑﺳﺗﮫ ﺷده اﺳت‪) .‬ﻧﺗرﺳﯾد(‪.‬‬‫ﭘس از اﻧدازهﮔﯾری‬‫ﭘس از اﻧدازهﮔﯾری‪ ،‬دﮐﻣﮫ ‪ START‬را ﺑزﻧﯾد ﺗﺎ دﺳﺗﮕﺎه ﺧﺎﻣوش ﺷود‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑردارﯾد و اطﻼﻋﺎت را ﺛﺑت ﮐﻧﯾد‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﯾن دﺳﺗﮕﺎه دارای ﻋﻣﻠﮑرد ﺧﺎﻣوش ﮐردن ﺧودﮐﺎر اﺳت ﮐﮫ ﺣدود ﯾﮏ دﻗﯾﻘﮫ ﭘس از اﻧدازهﮔﯾری دﺳﺗﮕﺎه را‬‫ﺧﺎﻣوش ﻣﯽﮐﻧد‪.‬‬‫در ﺻورت ﻧﯾﺎز ﺑﮫ ﺗﮑرار اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد‪ ،‬ﺣداﻗل ﺳﮫ دﻗﯾﻘﮫ ﻣﯾﺎن ھر ﺑﺎر اﻧدازهﮔﯾری‬‫ﻓﺎﺻﻠﮫ ﺑﯾﻧدازﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪13‬‬‫اﻧدازهﮔﯾری‬‫ﭘﯾش از اﻧدازهﮔﯾری‪ ،‬ﺑﮫ "ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻧرﻣﺎل‬‫‪.1‬‬‫ﺑﺎزوﺑﻧد را دور ﺑﺎزو )ﺗرﺟﯾﺣﺎ ً روی ﺑﺎزوی ﭼپ( ﻗرار دھﯾد‪.‬‬‫در ﺣﯾن اﻧدازهﮔﯾری آرام ﺑﻣﺎﻧﯾد‪.‬‬‫‪.2‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ھﻣﮫی ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر ﻧﻣﺎﯾﺎن ﻣﯽﺷوﻧد‪.‬‬‫ﺻﻔر )‪ (0‬ﻧﻣﺎﯾﺎن ﻣﯽﺷود ﮐﮫ اﻧدﮐﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ اﻧدازهﮔﯾری آﻏﺎز ﻣﯽﺷود‪ ،‬ھﻣﺎنطور ﮐﮫ در‬‫ﺗﺻوﯾر ﺳﻣت راﺳت ﻧﺷﺎن داده ﺷده اﺳت‪ ،‬ﻧﻣﺎﯾﺷﮕر ﺗﻐﯾﯾر‬‫ﻣﯽﮐﻧد‪ .‬ﺑﺎزوﺑﻧد ﺷروع ﺑﮫ ﺑﺎد ﺷدن ﻣﯽﮐﻧد‪ .‬اﺣﺳﺎس ﺳﻔﺗﯽ‬‫ﺑﺎزوﺑﻧد طﺑﯾﻌﯽ اﺳت‪ .‬ﺷﺎﺧص ﺑﺎر ﻓﺷﺎر در ﻟﺑﮫ ﺳﻣت ﭼپ‬‫ﻧﻣﺎﯾﺷﮕر در طول ﺑﺎد ﮐردن ﻧﻣﺎﯾﺎن ﻣﯽﺷود‪.‬‬‫ﻧﮑﺗﮫ‪ :‬ھر زﻣﺎن ﺧواﺳﺗﯾد ﺑﺎد ﮐردن را ﻣﺗوﻗف ﮐﻧﯾد‪،‬‬‫دﮐﻣﮫ ‪ START‬را ﻣﺟدداً ﻓﺷﺎر دھﯾد‪.‬‬‫‪.3‬‬‫وﻗﺗﯽ ﺑﺎد ﮐردن ﺗﻣﺎم ﺷد‪ ،‬ﺧﺎرج ﺷدن ﺑﺎد ﺑﮫﺻورت‬‫)ﻋﻼﻣت ﻗﻠب( ﺷروع ﺑﮫ‬‫ﺧودﮐﺎر آﻏﺎز ﺷده و‬‫ﭼﺷﻣﮏ زدن ﻣﯽﮐﻧد‪ ،‬ﮐﮫ ﻧﺷﺎن ﻣﯽدھد اﻧدازهﮔﯾری در‬‫ﺣﺎل اﻧﺟﺎم اﺳت‪ .‬ھﻧﮕﺎﻣﯽ ﮐﮫ ﺿرﺑﺎن ﻗﻠب ﺗﺷﺧﯾص داده‬‫ﺷد‪ ،‬ﻋﻼﻣت ﻗﻠب ﺑﺎ ھر ﺿرﺑﺎن ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﻧﮑﺗﮫ‪ :‬اﮔر ﻓﺷﺎر ﻣﻧﺎﺳب ﺑﮫ دﺳت ﻧﯾﺎﻣد‪ ،‬دﺳﺗﮕﺎه ﺑﮫطور‬‫ﺧودﮐﺎر ﻣﺟدداً ﺷروع ﺑﮫ ﺑﺎد ﺷدن ﻣﯽﮐﻧد‪.‬‬‫ﺑرای اﺟﺗﻧﺎب از ﺑﺎد ﺷدن ﻣﺟدد‪ ،‬ﺑﮫ "اﻧدازهﮔﯾری ﺑﺎ‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫‪.4‬‬‫وﻗﺗﯽ اﻧدازهﮔﯾری ﺗﻣﺎم ﺷد‪ ،‬ﻗراﺋت ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ و‬‫دﯾﺎﺳﺗوﻟﯽ و ﺿرﺑﺎن ﻗﻠب ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‪.‬‬‫ھوای ﺑﺎﻗﯽﻣﺎﻧده از ﺑﺎزوﺑﻧد ﺧﺎرج ﺷده و ﺑﺎد آن ﺑﮫطور‬‫ﮐﺎﻣل ﺧﺎﻟﯽ ﻣﯽﺷود‪.‬‬‫‪.5‬‬‫ﺑرای ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه‪ ،‬دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ﭘس از ﯾﮏ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺧودش ﺑﮫطور ﺧودﮐﺎر‬‫ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد را ﺑﺎ ﻓﺎﺻﻠﮫ ﺣداﻗل‬‫ﺳﮫ دﻗﯾﻘﮫ اﻧﺟﺎم دھﯾد‪.‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫‪START‬‬‫ﻓﺷﺎر دھﯾد‬‫ﮐﻠﯾﮫ ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر‬‫ﻧﻣﺎﯾﺎن ﻣﯽﺷوﻧد‬‫ﻧﻣﺎﯾش ﺻﻔر‬‫ﺷروع ﺑﺎد ﮐردن‬‫ﺗﻧظﯾم ﻓﺷﺎر‬‫اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ ‪WHO‬‬‫ﺿرﺑﺎن ﻗﻠب‬‫ﻓﺎرﺳﯽ ‪14‬‬‫ھوای ﺑﺎﻗﯽﻣﺎﻧده را ﺑﮫﺻورت ﺧودﮐﺎر‬‫ﺑﯾرون ﻣﯽدھد‬‫اﻧدازهﮔﯾری‬‫اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫‪ UA-767S-W‬ﺑرای ﺷﻧﺎﺳﺎﯾﯽ ﺿرﺑﺎن ﻗﻠب و ﺑﺎد ﮐردن ﺧودﮐﺎر ﺑﺎزوﺑﻧد ﺗﺎ رﺳﯾدن ﺑﮫ ﺳطﺢ ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ ﻣﺷﺧص طراﺣﯽ ﺷده اﺳت‪.‬‬‫از اﯾن روش ھﻧﮕﺎﻣﯽ اﺳﺗﻔﺎده ﮐﻧﯾد ﮐﮫ ﺑﺎد ﮐردن ﻣﺟدد ﭼﻧدﯾن ﺑﺎر ﺗﮑرار ﻣﯽﺷود ﯾﺎ ھﻧﮕﺎﻣﯽ ﮐﮫ ﺣﺗﯽ ﺑﺎ‬‫ﮐﺎھش ﻓﺷﺎر ﺑﮫ ‪ ۲۰‬ﻣﯾﻠﯽﻟﯾﺗر ﺟﯾوه ﯾﺎ ﮐﻣﺗر‪ ،‬ﻧﺗﺎﯾﺞ ﻧﺷﺎن داده ﻧﻣﯽﺷوﻧد‪.‬‬‫‪.1‬‬‫ﺑﺎزوﺑﻧد را دور ﺑﺎزو )ﺗرﺟﯾﺣﺎ ً روی ﺑﺎزوی ﭼپ( و‬‫ھمﺗراز ﺑﺎ ﻗﻠب ﺑﺑﻧدﯾد‪.‬‬‫‪.2‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫‪.3‬‬‫وﻗﺗﯽ ﻋﻼﻣت ﺻﻔر ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐرد‪،‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر داده و ﻧﮕﺎه دارﯾد ﺗﺎ رﻗﻣﯽ‬‫در ﺣدود ‪ ۳۰‬ﺗﺎ ‪ ۴۰‬ﻣﯾﻠﯽﮔرم ﺟﯾوه ﺑﺎﻻﺗر از ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ ﻣورد ﻧظر ظﺎھر ﺷود‪.‬‬‫‪.4‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﮫ رﻗم ﻣورد ﻧظر رﺳﯾدﯾد‪ ،‬دﮐﻣﮫ‬‫‪ START‬را رھﺎ ﮐﻧﯾد ﺗﺎ اﻧدازهﮔﯾری آﻏﺎز ﺷود‪ .‬ﺑﮫ‬‫ھﻣﺎن ﺗرﺗﯾﺑﯽ ﮐﮫ در ﺻﻔﺣﮫ ﻗﺑل ﺗوﺿﯾﺢ داده ﺷد‪ ،‬ﺑﮫ‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧوﻧﺗﺎن اداﻣﮫ دھﯾد‪.‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫‪ START‬دﮐﻣﮫ‬‫را ﻓﺷﺎر دھﯾد‬‫دﮐﻣﮫ را در ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫رھﺎ ﮐﻧﯾد‬‫‪START‬‬‫دﮐﻣﮫ را ﻓﺷﺎر داده و ﻧﮕﮫ دارﯾد‬‫ﺑرای ﻣﺷﺎھده ﺟزﺋﯾﺎت ﺑﯾﺷﺗر در ﻣورد اﻧدازهﮔﯾری ﺑﮫ‬‫ﺻﻔﺣﮫ ﻗﺑل ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ‬‫در ﻣوﻗﻌﯾت راﺣﺗﯽ ﺑﻧﺷﯾﻧﯾد‪ .‬دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﻗرار دھﯾد ﺑﮫ ﮔوﻧﮫای ﮐﮫ ﮐف دﺳتھﺎﯾﺗﺎن رو ﺑﮫ‬‫ﺑﺎﻻ ﺑوده و ﺑﺎزوﺑﻧد و ﻗﻠﺑﺗﺎن ھمﺗراز ﺑﺎﺷﻧد‪.‬‬‫ﭘﯾش از آﻏﺎز اﻧدازهﮔﯾری‪ ،‬ﺣدود ﭘﻧﺞ ﺗﺎ ده دﻗﯾﻘﮫ آرام ﺑﺎﺷﯾد‪ .‬اﮔر ﺑﮫ دﻟﯾل ﺗﻧشھﺎی ھﯾﺟﺎﻧﯽ دﭼﺎر ھﯾﺟﺎن ﯾﺎ‬‫اﻓﺳردﮔﯽ ھﺳﺗﯾد‪ ،‬ﻓﺷﺎر ﺧوﻧﺗﺎن ﺑﺎﻻﺗر )ﯾﺎ ﭘﺎﯾﯾنﺗر( از ﻓﺷﺎر ﺧون ﻧرﻣﺎل ﻧﺷﺎن داده ﺧواھد ﺷد‪ ،‬و ﺿرﺑﺎن‬‫ﻗﻠب ﻧﯾز ﻣﻌﻣوﻻً ﺳرﯾﻊﺗر از ﺣﺎﻟت ﻧرﻣﺎل ﻣﯽﺑﺎﺷد‪.‬‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد ﻣرﺗﺑﺎ ً در ﺣﺎل ﺗﻐﯾﯾر ﺑوده و واﺑﺳﺗﮫ ﺑﮫ ﮐﺎری ﮐﮫ اﻧﺟﺎم ﻣﯽدھد و ﻏذاﯾﯽ ﮐﮫ ﺧورده‬‫ﻣﯽﺑﺎﺷد‪ .‬آﻧﭼﮫ ﻧوﺷﯾدهاﯾد ﻣﯽﺗواﻧد ﺗﺄﺛﯾر ﺑﺳﯾﺎر ﻗوی و ﺳرﯾﻌﯽ روی ﻓﺷﺎر ﺧوﻧﺗﺎن داﺷﺗﮫ ﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﻣﺑﻧﺎی اﻧدازهﮔﯾریھﺎﯾش را روی ﺿرﺑﺎن ﻗﻠب ﻣﯽﮔذارد‪ .‬اﮔر ﺿرﺑﺎن ﻗﻠب ﺑﺳﯾﺎر ﺿﻌﯾف ﯾﺎ‬‫ﻧﺎﻣﻧظﻣﯽ داﺷﺗﮫ ﺑﺎﺷﯾد‪ ،‬ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫﺳﺧﺗﯽ ﺑﺗواﻧد ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﺷﺧﯾص دھد‪.‬‬‫اﮔر دﺳﺗﮕﺎه وﺟود ﺷراﯾطﯽ را ﺗﺷﺧﯾص دھد ﮐﮫ ﻏﯾر ﻧرﻣﺎل اﺳت‪ ،‬اﻧدازهﮔﯾری را ﻣﺗوﻗف ﮐرده و ﻧﻣﺎد ﺧطﺎ ظﺎھر‬‫ﻣﯽﺷود‪ .‬ﺑرای ﻣﺷﺎھده ﺗوﺿﯾﺢ ﻧﻣﺎدھﺎ‪ ،‬ﺑﮫ ﺻﻔﺣﮫ ‪ 7‬ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪15‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﺳﺗﻔﺎده ﺑزرﮔﺳﺎﻻن طراﺣﯽ ﺷده اﺳت‪ .‬اﮔر ﻣﯽﺧواھﯾد از اﯾن دﺳﺗﮕﺎه ﺑرای ﮐودﮐﺎن‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪ .‬ﮐودﮐﺎن ﻧﺑﺎﯾد ﺑدون ﮐﻣﮏ ﺑزرﮔﺳﺎﻻن از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﻧد‪.‬‬‫ﻋﻣﻠﮑرد ﺧودﮐﺎر دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﻣﯽﺗواﻧد ﺗﺣت ﺗﺄﺛﯾر دﻣﺎ ﯾﺎ رطوﺑت ﺷدﯾد ﯾﺎ ارﺗﻔﺎع ﻗرار ﮔﯾرد‪.‬‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ‬‫ﻧﮑﺗﮫ‪:‬‬‫اﯾن دﺳﺗﮕﺎه ‪ ۶۰‬اﻧدازهﮔﯾری آﺧر را در ﺣﺎﻓظﮫاش ذﺧﯾره ﻣﯽﮐﻧد‪.‬‬‫‪.1‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ھﯾﭻ ﭼﯾزی ﻧﻣﺎﯾش داده ﻧﻣﯽﺷود‪،‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد ﺗﺎ اطﻼﻋﺎت‬‫ذﺧﯾرهﺷده ﻓراﺧواﻧده ﺷود‪.‬‬‫‪.2‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت ﻧﺷﺎن داده ﺷد‪ ،‬دﮐﻣﮫ‬‫را رھﺎ ﮐﻧﯾد‪.‬‬‫‪.3‬‬‫ﺷﻣﺎره اطﻼﻋﺎت و اطﻼﻋﺎت ذﺧﯾره ﺷده ﺑﮫﺻورت‬‫ﺧودﮐﺎر و ﺑﮫ ﺗرﺗﯾب از آﺧرﯾن اﻧدازهﮔﯾری ﻧﺷﺎن‬‫داده ﻣﯽﺷوﻧد‪.‬‬‫‪.4‬‬‫ﭘس از آن ﮐﮫ ﺗﻣﺎﻣﯽ اطﻼﻋﺎت ﻧﻣﺎﯾش داده ﺷد‪،‬‬‫ﻧﻣﺎﯾﺷﮕر ﺧﺎﻣوش ﻣﯽﺷود‪.‬‬‫دﮐﻣﮫ ‪ START‬ﻓﺷﺎر دھﯾد و‬‫ﻧﮕﮫ دارﯾد‬‫ﻣﯾﺎﻧﮕﯾن ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن ﺿرﺑﺎن‬‫ﻧﮑﺗﮫ‪:‬‬‫اﮔر دﮐﻣﮫ ‪ START‬را در زﻣﺎن ﻓراﺧواﻧﯽ‬‫اطﻼﻋﺎت ﻓﺷﺎر دھﯾد‪ ،‬دﺳﺗﮕﺎه ﺧﺎﻣوش ﻣﯽﺷود‪.‬‬‫ﻣﯾﺎﻧﮕﯾن‬‫ھﻣﮫ اطﻼﻋﺎت‬‫ﺳﯾﺳﺗوﻟﯽ‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﺿرﺑﺎن‬‫ﺟدﯾدﺗرﯾن اطﻼﻋﺎت‬‫ﺳﯾﺳﺗوﻟﯽ‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﺿرﺑﺎن‬‫آﺧرﯾن اطﻼﻋﺎت )ﻗدﯾﻣﯽﺗرﯾن(‬‫ﻓﺎرﺳﯽ ‪16‬‬‫ﺷﺎﺧص ‪ IHB/AFib‬ﭼﯾﺳت؟‬‫وﻗﺗﯽ دﺳﺗﮕﺎه رﯾﺗم ﻧﺎﻣﻧظﻣﯽ را در طول اﻧدازهﮔﯾری ﺷﻧﺎﺳﺎﯾﯽ ﮐﻧد‪ ،‬ﺷﺎﺧص ‪ IHB/AFib‬روی ﻧﻣﺎﯾﺷﮕر ﺑﺎ‬‫ﻣﻘﺎدﯾر اﻧدازهﮔﯾری ظﺎھر ﻣﯽﺷود‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫ﻓرﮐﺎﻧس ﺷﺎﺧص ‪ ،IHB/AFib‬ﺑﺎ ﭘزﺷﮑﺗﺎن‬‫ﺗوﺻﯾﮫ ﻣﯽﺷود ﮐﮫ در ﺻورت ﻣﺷﺎھده اﯾن‬‫ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫‪ AFib‬ﭼﯾﺳت؟‬‫اﻧﻘﺑﺎﺿﺎت ﻗﻠب ﺑﮫ دﻟﯾل ﺳﯾﮕﻧﺎلھﺎی اﻟﮑﺗرﯾﮑﯽ ﮐﮫ در ﻗﻠب روی داده و ﺧون را ﺑﮫ ﮐل ﺑدن ﻣﯽﻓرﺳﺗد‪.‬‬‫ﻓﯾﺑرﯾﻼﺳﯾون دھﻠﯾزی )‪ (AFib‬زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ ﺳﯾﮕﻧﺎل اﻟﮑﺗرﯾﮑﯽ در دھﻠﯾز ﻗﻠب ﻣﻐﺷوش ﺷده و‬‫ﻣﻧﺟر ﺑﮫ آﺷﻔﺗﮕﯽ در ﻓواﺻل ﺿرﺑﺎن ﻣﯽﺷود‪ AFib .‬ﻣﯽﺗواﻧد ﻣوﺟب ﺷود ﮐﮫ ﺧون در ﻗﻠب از ﺣرﮐت‬‫ﺑﺎزاﯾﺳﺗد ﮐﮫ ﺑﮫﺳﺎدﮔﯽ ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﻟﺧﺗﮫ ﺧون و ﺑﮫ ﺗﺑﻊ آن ﻣوﺟب ﺳﮑﺗﮫ و ﺣﻣﻠﮫ ﻗﻠﺑﯽ ﺷود‪.‬‬‫‪%IHB/AFib‬‬‫‪ %IHB/AFib‬ﺑﮫﺻورت ﻓرﮐﺎﻧس ‪ IHB/AFib‬ﺷﻧﺎﺳﺎﯾﯽ ﺷده ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫‪ IHB/AFib‬ﻧﮫ ﺗﻧﮭﺎ ﻣﯽﺗواﻧد ﻧوﯾزھﺎﯾﯽ ﻣﺎﻧﻧد ﺣرﮐت ﻓﯾزﯾﮑﯽ را ﺗﺷﺧﯾص دھد‪ ،‬ﺑﻠﮑﮫ ﻗﺎدر اﺳت ﺿرﺑﺎن‬‫ﻗﻠب ﻧﺎﻣﻧظم را ﻧﯾز ﺷﻧﺎﺳﺎﯾﯽ ﻧﻣﺎﯾد‪ .‬در ﻧﺗﯾﺟﮫ‪ ،‬ﺗوﺻﯾﮫ ﻣﯽﮐﻧﯾم ﮐﮫ اﮔر ﻣﯾزان ‪ %IHB/AFib‬ﺑﺎﻻﺳت‪ ،‬ﺑﺎ‬‫ﭘزﺷﮑﺗﺎن ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫ﺗﻌداد ‪IHB/AFibs‬‬‫ﺷﻧﺎﺳﺎﯾﯽﺷده در ﺣﺎﻓظﮫ‬‫]‪x 100 [%‬‬‫= ‪%IHB/AFib‬‬‫ﺗﻌداد ﮐل‬‫ﻧﻣﺎﯾش ‪%IHB/AFib:‬‬‫‪ %IHB/AFib‬زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ ﻣﻘﺎدﯾر ﻣﯾﺎﻧﮕﯾن ﻧﺷﺎن داده ﻣﯽﺷوﻧد‪) .‬ﺑﮫ "‪.2‬‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت" در "ﺣﺎﻟت ﻋﻣﻠﯾﺎت" ﻣراﺟﻌﮫ ﮐﻧﯾد(‬‫‪ %IHB/AFib‬زﻣﺎﻧﯽ ﮐﮫ ﺷﻣﺎره ﺣﺎﻓظﮫ ﺷش ﯾﺎ ﮐﻣﺗر اﺳت‪ ،‬ﻧﻣﺎﯾش داده ﻧﻣﯽﺷود‪.‬‬‫ﻧﻣﺎﯾش ﻣﻘدار ﻣﯾﺎﻧﮕﯾن‬‫‪Level 0‬‬‫‪IHB/AFib=0%‬‬‫‪Level 3‬‬‫‪IHB/AFib=25 - 100%‬‬‫‪Level 1‬‬‫‪IHB/AFib=1 - 9%‬‬‫‪Level 2‬‬‫‪IHB/AFib=10 - 24%‬‬‫ﻋدم ﻧﻣﺎﯾش‬‫ﻓﺎرﺳﯽ ‪17‬‬‫ﺷﺎﺧص ﺑﺎر ﻓﺷﺎر‬‫اﯾن ﺷﺎﺧص ﭘﯾﺷروی ﻓﺷﺎر در طول اﻧدازهﮔﯾری را ﭘﺎﯾش ﻣﯽﮐﻧد‪.‬‬‫ﺗﻧظﯾم ﻓﺷﺎر‬‫آزاد ﺷدن ھوا‬‫اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‬‫ﺑﺎد ﮐردن در ﺣﺎل اﻧﺟﺎم اﺳت‬‫اﺗﻣﺎم ﺑﺎد ﮐردن‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫ھر ﺑﺧش ﺷﺎﺧص ﺑﺎر ﻣﺗﻧﺎﺳب ﺑﺎ طﺑﻘﮫﺑﻧدی ﻓﺷﺎر ﺧون ‪WHO‬‬‫اﺳت ﮐﮫ در ﺻﻔﺣﮫ ﺑﻌد ﺗوﺻﯾف ﻣﯽﺷود‪.‬‬‫ﻣﺛﺎل‪:‬‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ‪WHO‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻی ﺷدﯾد‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻی ﻣﺗوﺳط‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻی ﺧﻔﯾف‬‫ﻧرﻣﺎل ﺑﺎﻻ‬‫ﻧرﻣﺎل‬‫ﺑﮭﯾﻧﮫ‬‫‪ :‬اﯾن ﺷﺎﺧص ﯾﮏ ﺑﺧش را ﻧﻣﺎﯾش ﻣﯽدھد ﮐﮫ ﺑر ﻣﺑﻧﺎی‬‫اطﻼﻋﺎت ﮐﻧوﻧﯽ و ﻣﺗﻧﺎﺳب ﺑﺎ طﺑﻘﮫﺑﻧدی ‪ WHO‬ﻣﯽﺑﺎﺷد‪.‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻی ﻣﺗوﺳط‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻی ﺧﻔﯾف‬‫ﻧرﻣﺎل ﺑﺎﻻ‬‫در ﻣورد ﻓﺷﺎر ﺧون‬‫ﻓﺷﺎر ﺧون ﭼﯾﺳت؟‬‫ﻓﺷﺎر ﺧون ﻋﺑﺎرت از ﻧﯾروی وارد ﺷده ﺗوﺳط ﺧون ﺑﮫ دﯾوارهھﺎی رگ ﻣﯽﺑﺎﺷد‪ .‬ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ زﻣﺎﻧﯽ روی‬‫ﻣﯽدھد ﮐﮫ ﻗﻠب ﻣﻧﻘﺑض ﻣﯽﺷود‪ .‬ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ ﻗﻠب ﻣﻧﺑﺳط ﻣﯽﺷود‪ .‬ﻓﺷﺎر ﺧون ﺑر‬‫ﺣﺳب ﻣﯾﻠﯽﻣﺗر ﺟﯾوه )‪ (mmHg‬اﻧدازهﮔﯾری ﻣﯽﺷود‪ .‬ﻓﺷﺎر ﺧون طﺑﯾﻌﯽ ﯾﮏ ﻓرد ﺗوﺳط ﻓﺷﺎر ﭘﺎﯾﮫ ﻧﺷﺎن داده‬‫ﻣﯽﺷود ﮐﮫ اول ﺻﺑﺢ زﻣﺎﻧﯽ ﮐﮫ ﻓرد ھﻧوز در ﺣﺎل اﺳﺗراﺣت اﺳت و ﻏذاﯾﯽ ﻧﺧورده اﻧدازهﮔﯾری ﻣﯽﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪18‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﭼﯾﺳت و ﭼﮕوﻧﮫ ﮐﻧﺗرل ﻣﯽﺷود؟‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﮐﮫ ﺑﮫ ﻣﻌﻧﺎی ﻓﺷﺎر ﺧون ﻏﯾر ﻧرﻣﺎل ﺑﺎﻻی ﺷرﯾﺎﻧﯽ اﺳت‪ ،‬درﺻورﺗﯽﮐﮫ درﻣﺎن ﻧﺷود‬‫ﻣﯽﺗواﻧد ﻣﺷﮑﻼت زﯾﺎدی را ﺑرای ﻗﻠب از ﺟﻣﻠﮫ ﺳﮑﺗﮫ و ﺣﻣﻠﮫ ﻗﻠﺑﯽ اﯾﺟﺎد ﮐﻧد‪ .‬اﻓزاﯾش ﻓﺷﺎر ﺧون ﻣﯽﺗواﻧد‬‫از طرﯾﻖ ﺗﻐﯾﯾر ﺳﺑﮏ زﻧدﮔﯽ‪ ،‬ﭘرھﯾز از اﺳﺗرس و ﻣﺻرف دارو ﺗﺣت ﻧظر ﭘزﺷﮏ ﮐﻧﺗرل ﺷود‪.‬‬‫ﺑرای ﭘﯾﺷﮕﯾری از اﺑﺗﻼ ﺑﮫ اﻓزاﯾش ﻓﺷﺎر ﺧون‪ ،‬ﺑﺎﯾد آن را ﺗﺣت ﮐﻧﺗرل داﺷﺗﮫ ﺑﺎﺷﯾد‪:‬‬‫ﻣرﺗب ورزش ﮐﻧﯾد‬‫ﺳﯾﮕﺎر ﻧﮑﺷﯾد‬‫ﺑﮫطور ﻣرﺗب ﺑرای ﭼﮏ آپ و ﻣﻌﺎﯾﻧﮫ ﭘزﺷﮑﯽ‬‫ﻣﺻرف ﻧﻣﮏ و ﭼرﺑﯽ را ﮐﺎھش دھﯾد‬‫ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫وزن ﻣﻧﺎﺳب داﺷﺗﮫ ﺑﺎﺷﯾد‬‫ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ اﻧدازهﮔﯾری ﮐﻧم؟‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون در ﮐﻠﯾﻧﯾﮏ ﯾﺎ ﻣطب ﭘزﺷﮏ ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺗرس در ﻓرد ﺷده و اﯾن اﻣر‬‫ﻣﯽﺗواﻧد ﺑﺎﻋث اﻓزاﯾش ﻗراﺋت دﺳﺗﮕﺎه ﺑﮫ ﻣﯾزان ‪ ۲۵‬ﺗﺎ ‪ ۳۰‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه ﺑﯾﺷﺗر از ﻓﺷﺎر ﺧون اﻧدازهﮔﯾری‬‫ﺷده در ﻣﻧزل ﺷود‪ .‬اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون در ﺧﺎﻧﮫ ﻣوﺟب ﮐﺎھش ﺗﺄﺛﯾر ﻋواﻣل ﺧﺎرﺟﯽ ﺑر ﻗراﺋت ﻓﺷﺎر‬‫ﺧون ﻣﯽﺷود‪ ،‬ﻣﮑﻣل ﻗراﺋت ﭘزﺷﮏ اﺳت و ﺳﺎﺑﻘﮫ دﻗﯾﻖﺗر و ﮐﺎﻣلﺗری از ﻓﺷﺎر ﺧوﻧﺗﺎن را در اﺧﺗﯾﺎر ﭘزﺷﮏ‬‫ﻗرار ﻣﯽدھد‪.‬‬‫طﺑﻘﮫﺑﻧدی ﻓﺷﺎر ﺧون ‪WHO‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ درﺟﮫ ‪) ۳‬ﺷدﯾد(‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ درﺟﮫ ‪) ۲‬ﻣﺗوﺳط(‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ درﺟﮫ ‪) ۱‬ﺧﻔﯾف(‬‫ﻧوﺳﺎﻧﺎت ﻓﺷﺎر ﺧون‬‫ﻧرﻣﺎل ﺑﺎﻻ‬‫ﻓﺷﺎر ﺧون دﯾﺎﺳﺗوﻟﯽ‬‫اﺳﺗﺎﻧداردھﺎی ارزﯾﺎﺑﯽ ﻓﺷﺎر ﺧون ﺑﺎﻻ‪،‬‬‫ﺑدون در ﻧظر ﮔرﻓﺗن ﺳن‪ ،‬ﺗوﺳط‬‫ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪ (WHO‬طﺑﻖ‬‫ﻧﻣودار‪ ،‬اراﺋﮫ ﺷدهاﻧد‪.‬‬‫اﺳﻧﺎد ﻣرﺟﻊ‪ ،‬ﻣﺟﻠﮫ ﻓﺷﺎر ﺧون ﺑﺎﻻ ‪ ،۱۹۹۹‬ﺟﻠد ‪ ۱۷‬ﺷﻣﺎره ‪۲‬‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد در طول روز و‬‫در ﻓﺻول ﻣﺧﺗﻠف ﺑﮫ ﺷدت ﺗﻐﯾﯾر‬‫ﻣﯽﮐﻧد‪ .‬ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد ﻣﯽﺗواﻧد‬‫ﺑر اﺛر ﺷراﯾط ﻣﺧﺗﻠف در طول روز‬‫ﻓﺷﺎر ﺧون ﺳﯾﺳﺗوﻟﯽ‬‫ﺑﮫ ﻣﯾزان ‪ ۳۰‬ﺗﺎ ‪ ۵۰‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﺗﻐﯾﯾر ﮐﻧد‪ .‬در اﻓرادی ﮐﮫ دﭼﺎر ﻓﺷﺎر ﺧون ﺑﺎﻻ ھﺳﺗﻧد‪ ،‬اﯾن ﻧوﺳﺎﻧﺎت ﺣﺗﯽ ﺑﯾﺷﺗر ﻧﯾز ﻣﯽﺑﺎﺷد‪ .‬ﻣﻌﻣوﻻً‬‫ﻓﺷﺎر ﺧون در ﻣﺣل ﮐﺎر ﯾﺎ در زﻣﺎن ﺑﺎزی اﻓزاﯾش ﯾﺎﻓﺗﮫ و در طول ﺧواب ﺑﮫ ﮐﻣﺗرﯾن ﻣﯾزان ﺧود‬‫ﻣﯽرﺳد‪ .‬ﻟذا ﺑﺎ ﻣﺷﺎھده ﻧﺗﺎﯾﺞ ﯾﮏﺑﺎر اﻧدازهﮔﯾری ﺑﯾش از ﺣد ﻧﮕران ﻧﺷوﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون را ھر روز در زﻣﺎن ﻣﺷﺧﺻﯽ‬‫اﻧﺟﺎم داده و از دﺳﺗوراﻟﻌﻣﻠﯽ ﮐﮫ در اﯾن راھﻧﻣﺎ ﺗوﺿﯾﺢ‬‫ﺳﯾﺳﺗوﻟﯽ‬‫داده ﺷده اﺳت اﺳﺗﻔﺎده ﮐﻧﯾد ﺗﺎ ﻓﺷﺎر ﺧون ﻧرﻣﺎﻟﺗﺎن را ﭘﯾدا‬‫ﮐﻧﯾد‪ .‬ﻗراﺋت ﻣﻧظم ﻣﯽﺗواﻧد ﺳﺎﺑﻘﮫ ﺟﺎﻣﻌﯽ از ﻓﺷﺎر ﺧوﻧﺗﺎن‬‫را در اﺧﺗﯾﺎرﺗﺎن ﻗرار دھد‪ .‬ھﻧﮕﺎم ﺛﺑت ﻓﺷﺎر ﺧوﻧﺗﺎن‪،‬‬‫ﺣﺗﻣﺎ ً ﺗﺎرﯾﺦ و ﺳﺎﻋت را ﻗﯾد ﮐﻧﯾد‪ .‬ﺑرای ﺗﻔﺳﯾر اطﻼﻋﺎت‬‫ﻓﺷﺎر ﺧوﻧﺗﺎن‪ ،‬ﺑﺎ ﭘزﺷﮏ ﻣﺷورت ﮐﻧﯾد‪.‬‬‫ﺧواب‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﻧرﻣﺎل‬‫ﺑﮭﯿﻨﮫ‬‫ﻓﺷﺎر ﺧون‬‫زﻣﺎن‬‫ﻓﺎرﺳﯽ ‪19‬‬‫ﻋﯾبﯾﺎﺑﯽ‬‫ﻣﺷﮑل‬‫دﻟﯾل اﺣﺗﻣﺎﻟﯽ‬‫ھﯾﭻ ﭼﯾزی ﻧﻣﺎﯾش داده ﺑﺎطریھﺎ ﺧﺎﻟﯽ ﺷدهاﻧد‪.‬‬‫ﻧﻣﯽﺷود‪ ،‬ﺣﺗﯽ زﻣﺎﻧﯽ‬‫ﮐﮫ دﺳﺗﮕﺎه روﺷن‬‫اﻧﺗﮭﺎی ﺑﺎطریھﺎ در ﻣوﻗﻌﯾت‬‫اﺳت‪.‬‬‫ﺻﺣﯾﺢ ﻗرار ﻧﮕرﻓﺗﮫاﻧد‪.‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫ ﺷده‬‫ھﻣﮫ ﺑﺎطریھﺎ را ﺑﺎ ﺑﺎطریھﺎی ﻧو ﺟﺎﯾﮕزﯾن‬‫ﮐﻧﯾد‪.‬‬‫ﺑﺎطریھﺎ را ﺧﺎرج ﮐرده و آﻧﮭﺎ را ﻣطﺎﺑﻖ ﺑﺎ‬‫ﻋﻼﻣت ﻣﺛﺑت و ﻣﻧﻔﯽ ﮐﮫ در ﻣﺣﻔظﮫ ﺑﺎطری‬‫ﻧوﺷﺗﮫ ﺷده اﺳت ﻗرار دھﯾد‪.‬‬‫وﻟﺗﺎژ ﺑﺎطری ﺧﯾﻠﯽ ﮐم اﺳت‪.‬‬‫)ﻋﻼﻣت ‪LOW‬‬‫ھﻣﮫ ﺑﺎطریھﺎ را ﺑﺎ ﺑﺎطریھﺎی ﻧو ﺟﺎﯾﮕزﯾن‬‫ﺑﺎزوﺑﻧد ﺑﺎد ﻧﻣﯽﺷود‪ (BATTERY .‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﮐﻧﯾد‪.‬‬‫اﮔر ﺑﺎطریھﺎ ﺑﮫطور ﮐﺎﻣل ﺧﺎﻟﯽ ﺷده‬‫ﺑﺎﺷﻧد‪ ،‬اﯾن ﻋﻼﻣت ظﺎھر ﻧﻣﯽﺷود‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫ درﺳﺗﯽ روی ﺑﺎزو ﻗرار‬‫ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ دور ﺑﺎزو ﻗرار دھﯾد‪.‬‬‫داده ﻧﺷده اﺳت‪.‬‬‫در ﺣﯾن اﻧدازهﮔﯾری ﺑﺎزو ﯾﺎ ﺑدﻧﺗﺎن را ﻣطﻣﺋن ﺷوﯾد ﮐﮫ ﺑدن ﺷﻣﺎ در ﺣﯾن اﻧدازهﮔﯾری‬‫ﮐﺎﻣﻼً ﺑﯽﺣرﮐت و آرام ﺑﺎﺷد‪.‬‬‫ﺗﮑﺎن دادهاﯾد‪.‬‬‫دﺳﺗﮕﺎه اﻧدازهﮔﯾری‬‫راﺣت و ﺑﯽﺣرﮐت ﺑﻧﺷﯾﻧﯾد‪.‬‬‫ﻧﻣﯽﮐﻧد‪.‬‬‫ﻗراﺋتھﺎ ﺧﯾﻠﯽ ﺑﺎﻻ ﯾﺎ ﺑﺎزوﺑﻧد در ﻣوﻗﻌﯾت ﺻﺣﯾﺢ ﻗرار دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﻗرار دھﯾد ﺑﮫ ﮔوﻧﮫای ﮐﮫ‬‫ﻧدارد‪.‬‬‫ﮐف دﺳتھﺎﯾﺗﺎن رو ﺑﮫ ﺑﺎﻻ ﺑوده و ﺑﺎزوﺑﻧد و ﻗﻠﺑﺗﺎن‬‫ﺧﯾﻠﯽ ﭘﺎﯾﯾن ھﺳﺗﻧد‪.‬‬‫ھمﺗراز ﺑﺎﺷﻧد‪.‬‬‫اﮔر ﺿرﺑﺎن ﻗﻠب ﺧﯾﻠﯽ ﺿﻌﯾف ﯾﺎ ﻧﺎﻣﻧظﻣﯽ‬‫داﺷﺗﮫ ﺑﺎﺷﯾد‪ ،‬ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫ ﺳﺧﺗﯽ‬‫ﺑﺗواﻧد ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﻌﯾﯾن ﮐﻧد‪.‬‬‫ﻋدد ﻧﺷﺎن داده ﺷده ﺗوﺳط دﺳﺗﮕﺎه‬‫ﺑﮫ "ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ اﻧدازهﮔﯾری‬‫ﻣﺗﻔﺎوت از ﻋددی اﺳت ﮐﮫ در‬‫ﮐﻧم؟" ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﮐﻠﯾﻧﯾﮏ ﯾﺎ ﻣطب ﭘزﺷﮏ‬‫اﻧدازهﮔﯾری ﺷده اﺳت‪.‬‬‫ﺳﺎﯾر ﻣوارد‬‫ﺑﺎطریھﺎ را ﺧﺎرج ﮐﻧﯾد‪ .‬آﻧﮭﺎ را ﺑﮫ درﺳﺗﯽ‬‫ﻗرار دھﯾد و ﺳﻌﯽ ﮐﻧﯾد ﺑﺎر دﯾﮕر اﻧدازهﮔﯾری‬‫را اﻧﺟﺎم دھﯾد‪.‬‬‫اﮔر ﮐﻠﯾﮫ اﻗداﻣﺎت ذﮐرﺷده را اﻧﺟﺎم دادﯾد اﻣﺎ ﻣﺷﮑﻠﺗﺎن ﺣل ﻧﺷد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬ﺳﻌﯽ ﻧﮑﻧﯾد‬‫ﻧﮑﺗﮫ‪:‬‬‫اﯾن ﻣﺣﺻول را ﺑﺎز ﯾﺎ ﺗﻌﻣﯾر ﮐﻧﯾد‪ ،‬زﯾرا اﯾن ﮐﺎر ﻣوﺟب ﺑﯽاﻋﺗﺑﺎر ﺷدن ﮔﺎراﻧﺗﯽ آن ﻣﯽﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪20‬‬‫ﻧﮕﮭداری‬‫دﺳﺗﮕﺎه را ﺑﺎز ﻧﮑﻧﯾد‪ .‬اﯾن دﺳﺗﮕﺎه از اﺟزای اﻟﮑﺗرﯾﮑﯽ ظرﯾف و واﺣد ھوای ﭘﯾﭼﯾدهای ﺗﺷﮑﯾل ﺷده اﺳت ﮐﮫ‬‫ﻣﯽﺗواﻧد آﺳﯾب ﺑﺑﯾﻧد‪ .‬اﮔر ﻧﺗواﻧﺳﺗﯾد ﺑﺎ اﺳﺗﻔﺎده از راھﻧﻣﺎی ﻋﯾبﯾﺎﺑﯽ ﻣﺷﮑل دﺳﺗﮕﺎه را ﺣل ﮐﻧﯾد‪ ،‬ﺑﺎ ﻓروﺷﻧده‬‫ﻣﺟﺎز ﺷﮭرﺗﺎن ﯾﺎ ﮔروه ﺧدﻣﺎت ﻣﺷﺗرﯾﺎن ﻣﺎ ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬اطﻼﻋﺎت ﻓﻧﯽ‪ ،‬ﻗطﻌﺎت و واﺣدھﺎی ﯾدﮐﯽ از‬‫طرﯾﻖ ﺑﺧش ﺧدﻣﺎت ﻣﺷﺗرﯾﺎن ‪ A&D‬در اﺧﺗﯾﺎر ﻓروﺷﻧدﮔﺎن ﻣﺟﺎز ﻗرار ﻣﯽﮔﯾرﻧد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای ﺧدﻣﺎتدھﯽ طوﻻﻧﯽ ﻣدت طراﺣﯽ و ﺗوﻟﯾد ﺷده اﺳت‪ .‬اﻣﺎ ﺗوﺻﯾﮫ ﻣﯽﺷود ﮐﮫ ھر ‪ 2‬ﺳﺎل‬‫ﯾﮏﺑﺎر دﺳﺗﮕﺎه را ﺑررﺳﯽ ﮐﻧﯾد ﺗﺎ از ﮐﺎرﮐرد ﺻﺣﯾﺢ و دﻗت آن ﻣطﻣﺋن ﺷوﯾد‪ .‬ﻟطﻔﺎ ً ﺑرای ﺳروﯾس و‬‫ﻧﮕﮭداری دﺳﺗﮕﺎه‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ ﻣﺟﺎز ﯾﺎ ‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫اطﻼﻋﺎت ﻓﻧﯽ‬‫ﻧوع‬‫روش اﻧدازهﮔﯾری‬‫داﻣﻧﮫ اﻧدازهﮔﯾری‬‫دﻗت اﻧدازهﮔﯾری‬‫ﻣﻧﺑﻊ ﺗﻐذﯾﮫ‬‫ﺗﻌداد اﻧدازهﮔﯾریھﺎ‬‫طﺑﻘﮫﺑﻧدی‬‫آزﻣﺎﯾش ﮐﻠﯾﻧﯾﮑﯽ‬‫‪EMD‬‬‫ﺣﺎﻓظﮫ‬‫ﺷراﯾط ﮐﺎر‬‫ﺷراﯾط ﺣﻣل و ﻧﻘل‬‫‪ /‬ﻧﮕﮭداری‬‫اﺑﻌﺎد‬‫وزن‬‫ﺣﻔﺎظت ورودی‬‫‪UA-767S-W‬‬‫اﻧدازهﮔﯾری ﻧوﺳﺎن ﺳﻧﺟﯽ‬‫‪ 0‬ﺗﺎ ‪ 299‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﻓﺷﺎر‪:‬‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‪ 60 :‬ﺗﺎ ‪ 279‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‪ 40 :‬ﺗﺎ ‪ 200‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﺿرﺑﺎن‪ 40 - 180 :‬ﺑﯾت ﺑر دﻗﯾﻘﮫ‬‫‪ ±3‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‬‫ﻓﺷﺎر‪:‬‬‫ﺿرﺑﺎن‪±5% :‬‬‫‪ 4‬ﺑﺎطری ‪ 1.5‬وﻟﺗﯽ )‪ R6P، LR6‬ﯾﺎ ‪ (AA‬ﯾﺎ‬‫آداﭘﺗور‪) (TB-233C) AC‬ﻣوﺟود ﻧﯾﺳت(‬‫ﺣدود ‪ 700‬ﻣرﺗﺑﮫ ‪) LR6‬ﺑﺎطریھﺎی ﻗﻠﯾﺎﯾﯽ(‬‫ﺣدود ‪ 200‬ﻣرﺗﺑﮫ ‪) R6P‬ﺑﺎطریھﺎی ﻣﻧﮕﻧزی(‬‫ﺑﺎ ﻣﻘدار ﻓﺷﺎر ‪ 180‬ﻣﯾﻠﯽﻣﺗر ﺟﯾوه‪ ،‬دﻣﺎی اﺗﺎق ‪ 23‬درﺟﮫ ﺳﺎﻧﺗﯽﮔراد‪.‬‬‫ﺗﺟﮭﯾزات ‪ EM‬ﺑﺎ ﺗوان داﺧﻠﯽ )ﺗﺄﻣﯾنﺷده ﺗوﺳط ﺑﺎطری( ‪ /‬ﮐﻼس ‪) ۲‬ﺗﺄﻣﯾنﺷده ﺗوﺳط‬‫آداﭘﺗور(‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت ﭘﯾوﺳﺗﮫ‬‫ﻣطﺎﺑﻖ اﺳﺗﺎﻧدارد ‪ISO81060-2：2013‬‬‫در ﺑررﺳﯽ اﻋﺗﺑﺎرﺳﻧﺟﯽ ﺑﺎﻟﯾﻧﯽ‪ ،‬از ‪ K5‬ﺟﮭت ﺗﻌﯾﯾن ﻓﺷﺎر ﺧون دﯾﺎﺳﺗوﻟﯽ ﺑرای‬‫‪ 85‬ﺳوژه اﺳﺗﻔﺎده ﺷد‪.‬‬‫‪IEC 60601-1-2: 2014‬‬‫‪ 60‬اﻧدازهﮔﯾری اﺧﯾر‬‫‪ +10‬ﺗﺎ ‪ +40‬درﺟﮫ ﺳﺎﻧﺗﯽﮔراد ‪ /‬رطوﺑت ﻧﺳﺑﯽ ‪ ۱۵‬ﺗﺎ ‪ ۸۵‬درﺻد ‪ ۸۰۰ /‬ﺗﺎ ‪۱۰۶۰‬‬‫ھﮑﺗوﭘﺎﺳﮑﺎل‬‫‪ -20‬ﺗﺎ ‪ +60‬درﺟﮫ ﺳﺎﻧﺗﯽﮔراد ‪ /‬رطوﺑت ﻧﺳﺑﯽ ‪ 10‬ﺗﺎ ‪ 700 / 95%‬ﺗﺎ ‪1060‬‬‫ھﮑﺗوﭘﺎﺳﮑﺎل‬‫ﺣدود ‪) ۱۴۰‬ﻋرض( در ‪) ۶۰‬طول( در ‪) ۱۰۵‬ﻋﻣﻖ( ﻣﯾﻠﯽﻣﺗر‬‫ﺣدود ‪ ۲۴۵‬ﮔرم‪ ،‬ﺑدون ﺑﺎطری‬‫دﺳﺗﮕﺎه‪IP20:‬‬‫ﻓﺎرﺳﯽ ‪21‬‬‫ﻗطﻌﮫ ﺑﮫﮐﺎررﻓﺗﮫ‬‫ﻋﻣر ﻣﻔﯾد‬‫ﻧوع ‪BF‬‬‫ﺑﺎزوﺑﻧد‬‫دﺳﺗﮕﺎه‪ ۵ :‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫ﺑﺎزوﺑﻧد‪ ۲ :‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫آداﭘﺗور ‪ ۵: AC‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫آداﭘﺗور ‪ AC‬ﯾدﮐﯽ‬‫ً‬‫آداﭘﺗور ﺑرای ﻣﺗﺻل ﮐردن دﺳﺗﮕﺎه ﺑﮫ ﻣﻧﺑﻊ ﺑرق ﺧﺎﻧﮕﯽ ﻣﯽﺑﺎﺷد‪ .‬ﻟطﻔﺎ ﺑرای ﺧرﯾد آن ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ‬‫‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬آداﭘﺗور ‪ AC‬ﺑﺎﯾد ﺑﮫﺻورت دورهای ﺑررﺳﯽ و ﺗﻌوﯾض ﺷود‪.‬‬‫ﻟطﻔﺎ ً ﺑرای ﺧرﯾد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ ‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫‪TB-233C‬‬‫آداﭘﺗور ‪ AC‬ﺑﺎﯾد ﺑﮫﺻورت دورهای ﺑررﺳﯽ و ﺗﻌوﯾض ﺷود‪.‬‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی آداﭘﺗور ‪ AC‬ﭼﺎپ ﺷده اﺳت‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﺻرﻓﺎ ً ﺑرای اﺳﺗﻔﺎده در ﻣﺣﯾطھﺎی ﺳرﺑﺳﺗﮫ‬‫دﺳﺗﮕﺎه ﮐﻼس دو‬‫ﻓﯾوز ﺣرارﺗﯽ‬‫ﻓﯾوز‬‫ﺑرﭼﺳب دﺳﺗوراﻟﻌﻣل ‪ EC‬دﺳﺗﮕﺎه‬‫ﺑرﭼﺳب ﻣﺟوز ‪ EAC‬دﺳﺗﮕﺎه‬‫ﻗطبھﺎی ﻧری آداﭘﺗور ‪AC‬‬‫ﻟوازم ﻓرﻋﯽ ﮐﮫ ﺟدا ﻓروﺧﺗﮫ ﻣﯽﺷوﻧد‬‫اﻧدازه ﺑﺎزو‬‫اﻧدازه ﺑﺎزوﺑﻧد‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن ‪ ۳۱‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ ۴۵‬ﺳﺎﻧﺗﯽﻣﺗر‬‫‪CUF-F-LA‬‬‫‪ ۲۲‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ ۴۲‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫‪CUF-I‬‬‫‪ ۲۲‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ ۳۲‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪CUF-F-A‬‬‫اﻧدازه ﺑﺎزو‪ :‬ﻣﺣﯾط ﺑﺎزو در ﺑﺧش ﻋﺿﻠﮫ دو ﺳر‪.‬‬‫آداﭘﺗور ‪AC‬‬‫ﻧﮑﺗﮫ‪:‬‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪TB-233C‬‬‫دوﺷﺎﺧﮫ )ﻧوع ﭘرﯾز(‬‫ﻧوع ‪C‬‬‫ﻣﻣﮑن اﺳت ﻣﺷﺧﺻﺎت ﺑدون اطﻼع ﻗﺑﻠﯽ دﺳﺗﺧوش ﺗﻐﯾﯾراﺗﯽ ﺷوﻧد‪.‬‬‫طﺑﻘﮫﺑﻧدی ‪ IP‬ﻋﺑﺎرت از درﺟﮫ ﺣﻔﺎظت ﻓراھمﺷده ﺗوﺳط ﺿﻣﺎﯾم ﻣطﺎﺑﻖ ﺑﺎ ‪ IEC 60529‬ﻣﯽﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه در ﺑراﺑر اﺷﯾﺎی ﺧﺎرﺟﯽ ﺟﺎﻣد ﺑﺎ ﻗطر ‪ ۱۲‬ﻣﯾﻠﯽﻣﺗر و ﺑﯾﺷﺗر ﻣﺎﻧﻧد اﻧﮕﺷت ﻣﻘﺎوم ﻣﯽﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه در ﺑراﺑر آب ﻣﻘﺎوم ﻧﯾﺳت‪.‬‬‫ﻓﺎرﺳﯽ ‪22‬‬‫ﯾﺎدداﺷت‬‫ﻓﺎرﺳﯽ ‪23‬‬

A&D Company, Ltd.
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119

OX14 1DY

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409
OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司
中国 上海市浦东新区浦东大道138号永华大厦21楼A室 邮编200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599