SPECIFICATIONS

CarotidTransonic® Clinical Flowprobes
Models:
AU*, FMC, FME, FMV, MB/MR, MB-S/MR-S, MU

USA/CanadaM

Transonic Systems Inc.
34 Dutch Mill Road
Ithaca, NY 14850
U.S.A.
Tel: +1 607-257-5300
Fax: +1 607-257-7256
support@transonic.com

PTransonic Europe B.V.
Business Park Stein 205
6181 MB Elsloo
The Netherlands
Tel: +31 43-407-7200
Fax: +31 43-407-7201
europe@transonic.com

Asia/PacificJapan

Transonic Asia Inc.
6F-3 No 5 Hangsiang Rd.
Dayuan, Taoyuan County33747 Taiwan, R.O.C.
Tel: +886 3399-5806
Fax: +886 3399-5805
support@transonicasia.com

Nipro-Transonic Japan Inc.		
7th Floor, Maruha Building
11-1 Matsuba-cho
Tokorozawa City, Saitama
359-0044 Japan
Tel: +81 04-2946-8541
2797
Fax: +81 04-2946-8542
japan@transonic.com

AU-ISRT-ClinicalProbes-EN, Rev I Last Updated 10/31/23

CClinical Flowprobe Specifications
Probe Size

Vessel Size
Outer Diameter

mmmm

Catalog #Max Flow

Accuracy

Max Zero
OffsetUltrasoundFrequency

L/min% of Reading

mL/minMHz

FMC-Series: Coronary immediate-use steam sterilizable Probes (long handle, J-reflector)
1.5mm

1.5-2.3HQF1.5FMC

0.5+/- 10+/- 24.82mm

2.0-3.0

HQF2FMC0.5+/- 10+/- 24.83mm

2.7-4.0

HQF3FMC

1+/- 10+/- 3

2.44mm

3.0-5.0

HQF4FMC

2+/- 10+/- 6

2.4

FMV-Series: Vascular immediate-use steam sterilizable Probes (short handle, J-reflector)
1.5mm

1.5-2.3

HQF1.5FMV

0.5+/- 10+/- 2

4.82mm2.0-3.0

HQF2FMV

0.5+/- 10+/- 24.83mm

2.7-4.0

HQF3FMV

1+/- 10+/- 3

2.44mm

3.0-5.0HQF4FMV2+/- 10+/- 6

2.46mm

4.0-7.3

HQF6FMV5+/- 10+/- 15

1.88mm

6.4-9.8

HQF8FMV10+/- 10+/- 30

1.210mm8.0-12.0

HQF10FMV10+/- 10+/- 30

1.212mm

9.5-15.0

HQF12FMV

20+/- 10+/- 60

0.914mm

11.8-17.3

HQF14FMV

20+/- 10+/- 60

0.9

FME-Series: Carotid Endarterectomy immediate-use steam sterilizable Probes (short handle, L-reflector)
1.5mm

1.5-2.3

HQF1.5FME

0.5+/- 10+/- 24.82mm

2.0-3.0

HQF2FME

0.5+/- 10+/- 24.83mm

2.7-4.0

HQF3FME

1+/- 10+/- 32.44mm

3.0-5.0

HQF4FME

2+/- 10+/- 62.46mm

4.0-7.3

HQF6FME

5+/- 10+/- 15

1.88mm

6.4-9.8

HQF8FME

10+/- 10+/- 30

1.210mm8.0-12.0

HQF10FME

10+/- 10+/- 30

1.2MB-Series: Intracranial Charbel Micro-Flowprobes (long handle, single use)
1.5mm

1.1-1.6

HQG1.5MB

0.2+/- 10+/- 3

4.82mm

1.6-2.4

HQG2MB0.5+/- 10+/- 53.63mm

2.6-3.8

HQG3MB

1+/- 10+/- 10

2.4

MB-S-Series: Extracranial Micro-Flowprobes (short handle, single use)
3mm

2.7-4.0

HQG3MB-S

1+/- 10+/- 3

3.64mm

3.0-5.0

HQG4MB-S

2+/- 10+/- 6

2.46mm

4.0-7.3

HQG6MB-S5+/- 10+/- 15

1.8

MR-Series: Intracranial Charbel Micro-Flowprobes (long handle, multi-use)
1.5mm

1.1-1.6

HQE1.5MR

0.2+/- 10+/- 34.82mm

1.6-2.4

HQE2MR

0.5+/- 10+/- 5

3.63mm

2.6-3.8

HQE3MR

1+/- 10+/- 10

2.4

2AU-ISRT-ClinicalProbe-EN
Rev I

Clinical Flowprobe Specifications
Probe Size

Vessel Size
Outer Diameter

mmmm

Catalog #Max FlowAccuracy

Max ZeroOffset

UltrasoundFrequency

L/min% of Reading

mL/minMHz

MR-S-Series: Extracranial Micro-Flowprobes (short handle, multi-use)
3mm

2.7-4.0

HQE3MR-S

1+/- 10+/- 3

3.64mm

3.0-5.0

HQE4MR-S

2+/- 10+/- 6

2.46mm

4.0-7.3

HQE6MR-S5+/- 10+/- 15

1.8

MU-Series: Microvascular Probes (short handle, L-reflector)
0.7mm

0.5-0.73

HQF0.7MU

50 mL/min+/- 10+/- 0.4

9.61mm0.76-1.1

HQF1MU

100 mL/min+/- 10+/- 0.8

7.2

1.5mm

1.1-1.7

HQF1.5MU

200 mL/min+/- 10+/- 1.5

4.82mm

1.6-2.8

HQF2MU

500 mL/min+/- 10+/- 2.5

3.63mm

2.6-3.8

HQF3MU

1000 mL/min+/- 10+/- 5

2.4

AU-Series: COnfidence Flowprobes*
Size 8-36 available with HQF prefix; Size 4&6 available with HQE 8 use
4mm

3-4

HQE4AU2+/- 10+/- 67.26mm

5-6

HQE6AU5+/- 10+/- 84.88mm

6-8

HQF8AU10+/- 10+/- 30

3.610mm

8-10HQF10AU10+/- 10+/- 30

3.612mm

9-12HQF12AU

20+/- 10+/- 60

2.414mm

11-14

HQF14AU

20+/- 10+/- 60

2.416mm

13-16

HQF16AU

50+/- 10+/- 150

1.820mm15-18 w/ 18mm insert
17-20 w/ 20mm insert

HQF20AU

50+/- 10+/- 150

1.824mm18-22 w/ 22mm insert
20-24 w/ 24mm insert

HQF24AU

100+/- 10+/- 300

1.228mm22-26 w/ 26mm insert
24-28 w/ 28mm insert

HQF28AU

100+/- 10+/- 300

1.232mm26-30 w/ 30mm insert
28-32 w/ 32mm insert

HQF32AU

200+/- 10+/- 600

0.9

36mm

30-34 w/ 34mm insert
32-36 w/ 36mm insert

HQF36AU

200+/- 10+/- 600

0.9

HQE - 8 or 16 use
HQG - single use
HQF - 50 use*AU probes not available in the European Union

AU-ISRT-ClinicalProbe-EN
Rev I3