AD-4411-EIP/PRT/ECT
Weighing IndicatorPart names

Connection to power supply and connection to load cell

Calibration

Front panel

Attach the accessory power connector and wire as shown in the figure below.

Calibrate the AD-4411 to properly convert the signal from the load cell to a load value.
Please prepare a calibration weight.

(4)

Simplified Instruction Manual・ All rights reserved. No part of this manual may be reprinted or copied without theprior written permission of A&D.
・ The specifications and other information contained in this manual are subject tochange without notice in order to make improvements.
・ Every precaution has been taken in the preparation of this manual. Nevertheless,
A&D assumes no responsibility for errors or omissions. If you discover anyproblems with this manual, please notify A&D.
・ A&D Company, Limited shall not be liable for special, indirect, incidental, orconsequential damages, loss of profits or production or commercial loss in any way
connected with the products, whether such claim is based in contract, warranty,negligence, or strict liability.

(1)
(2)
(3)

No. Name(1) Main display(2)© 2023 A&D Company, Limited

ZERO status
NET status

The LED is ON when the net value is displayed.

STABLE status

The LED is ON when the measured value is stable.

S1 / S2 / S3status

The LED is ON when the S1 / S2 / S3 status ON condition(FncF07 / 08 / 09) is met.
Zeros the gross value. Moves the flashing digit to the leftwhen not in measurement mode.

(3)[TARE/↑] key

Performs tare. Increases the flashing digit by one when notin measurement mode.

[F1/↓] key

Performs the function set for the F1 key function (FncF05).
Decreases the flashing digit by one when not inmeasurement mode.

[F2/ ] key

Performs the function set for the F2 key function (FncF06).
Updates the setting value entered when not inmeasurement mode.

Telephone: [81] (3) 5391-6132 Fax: [81] (3) 5391-1566

Detailed instruction manualThis manual provides simplified precautions and operating instructions for AD-4411.
For further information about the AD-4411, please refer to the “AD-4411 Instruction
Manual” which is available for download from the A&D website (https://www.aandd.jp).

Introduction
The AD-4411 is a weighing indicator that can convert signals from strain gauge loadcells and connect them to an Ethernet-based field network. It contributes to an
efficient system by connecting weighing instruments to industrial control systems inplants and factories.
- Daisy-chain connection is possible without a switching hub, thanks to two
communication ports.
- 7-segment green LED display with a character height of 10mm and display
resolution of ±999999.
- High-speed AD conversion of 1200 times/second and digital filter enable highspeed and accuracy weighing.
- DIN96x48 panel mount type with IP65 protection on the front panel.
- PC can update the settings via USB port.

Safety precaution
If the equipment is used in a manner not specified by the manufacturer, the protection
provided by the equipment may be impaired.
Read the following precautions carefully before using the indicator for safe and correct
usage.

(4)

Capacity label

Changes the operation mode. Cancels the setting value
entered when not in measurement mode.
Attach the included capacity label, if necessary.

(5)

Unit label

Attach the included unit label, if necessary.

[MODE/ESC] key

CAUTION・ Do not clamp control wires or communication cables with power lines, or do notplace them close to power lines.
・ Place the load cell cable sufficiently away from high frequency circuits such as highvoltage power lines and inverter load circuit.
・ When the front cover have dirt, wipe them with wet soft cloth. Do not use organic
solvent such like benzine, thinner and alcohol. Doing so may result in deformation
or discoloration of the unit.
・ Suitable for use at pollution degree of 2 or less.
・ Use within an altitude of 0 to 2000m
・ To an external connection port connect the circuit separated from a dangerous
voltage by a double/reinforced insulation.

ItemSpecifications

Wire size

0.14 to 1.5 mm² (AWG 26 to 16)

Wire strip length

7 mmTightening torque0.22 to 0.25 Nm

After Power-On, press the [MODE/ESC] key morethan 3s.
24V

Press the [F1/↓] key twice.

0V

In the case of the 4-wire connection type, attach the accessory load cell connector
and wire as shown below. Change the load cell connection type (CALF17) in thecalibration function to 0: 4-wire type (default value = 1: 6-wire type).

(2)

(3)

Press the [F2/ ] key.

×2

Press the[F1/↓] key.

Actual load calibration
Press the [F2/ ] key.
ShieldPower supply (+)

Press the [F2/ ] key.

Signal (+)
Power supply (-)
Signal (-)

The current load cell input signal (mV/V) will be
displayed. Press the [F2/ ] key to execute Zerocalibration.
If Zero calibration is successful, "PASS" will be
displayed, and zero calibration will be completed.
Press the [F2/ ] key.

6-wire connection
Set load cell connection type (CALF17) to 1: 6 wire type (Default).
When you connect the load cells in parallel, use a summing box. Attach the accessory
load cell connector and wire as shown below.

Rear panel
Shield(1)

Power supply (+)
Signal (+)
Power supply (-)

Press the [F2/ ] key.
Set a calibration weight value by the following key
operations.
[ZERO/←] key: Moves the flashing digit to the left.
[TARE/↑] key: Increases the flashing digit by one.
[F1/↓] key: Decreases the flashing digit by one.
[F2/ ] key: Confirm the setting value.

Signal (-)

(4)

(5)
Summing boxNo.
(1)

Name
DC power input terminals(2)

Load cell input terminals(3)

USB connector(4)

Field network status LEDs

(5)

Field network connector

WARNING・ Provide an external safety circuit to the indicator so that the safety of the wholesystem can be secured even if errors occur in the external power supply or theindicator.
・ This indicator must be used indoors. Do not use the indicator in the following
environment:-where the temperature and the humidity exceed the specifications-where corrosive gases or flammable gases exist-where the indicator gets wet with oil, chemicals, or water-where the indicator is exposed to direct sunlight・ Turn off all the external power supplies used in the system before installing orremoving the indicator.
・ Turn off all the external power supplies used in the system before wiring.
・ Be sure to ground the indicator.

Applicable wire

Description
Displays measured value or various settings.
The LED is ON when the measured value is within 1/4 theminimum division.

[ZERO/←] key

1WMPD4004632C(5)

Description
Terminals for connection of a DC24V power
supply.
Terminals for connection of load cells.
Connector for connection with setting PC.
(Type-C)
Notifies field network status.
Connector for connection of PLC via fieldnetwork. Dual ports can be used for daisy chain
wiring (RJ-45).

Accessories
Waterproof packing, Panel mount bracket x2, Capacity label, Unit label,Power connector, Load cell connector.

Mounting to control panel
Insert the waterproof packing around the Unit, and insert the Unit through the front ofthe panel.
Insert the left and right mounting brackets into the case grooves and push until theyreach the panel.
Waterproof packing
Control panel
Panel mounting brackets

The current load cell input signal (mV/V) will be
displayed. Place the calibration weight or apply aload on the load cell. Press the [F2/ ] to execute
Span calibration.
If span calibration is successful, "PASS" will bedisplayed, and span calibration will be completed.
Press the [MODE/ESC] key four times to return tothe measurement mode.

×4

Function list

Specifications

Calibration function list

：Press to return to the previous screen.
Basic function

Item selection

Setting value input

For details, see the "AD-4411 Instruction Manual"on the A&D website (https://www.aandd.jp).

Default

01
02

Unit
Decimal point position

2003
04
05
06
07

Minimum division d
Maximum capacity
Zero setting range
Zero tracking time
Zero tracking width

08
09
10
1112

Stability detection time
Stability detection width
Zero-setting when unstable
Taring when unstable
Taring when the gross isnegative
Zero clear
Power-on zero
Condition of negative
overload
NTEP
Load cell connection type

0: None / 1: g / 2: kg / 3: t
0: 0 (No decimal point) / 1: 0.0 / 2: 0.00/ 3: 0.000 / 4: 0.0000 / 5: 0.00000
1: 1 d / 2: 2 d / 3: 5 d / 4: 10 d / 5: 20 d / 6: 50 d1 to 999999
0 to 100 %0.0 to 5.0 s
0: Disable / 1: 0.5 d / 2: 1.0 d / 3: 1.5 d / 4: 2.0 d/ 5: 2.5 d / 6: 3.0 d / 7: 3.5 d / 8: 4.0 d / 9: 4.5 d0.0 to 9.9 s
0 to 100 d
0: Disable / 1: Enable
0: Disable / 1: Enable
0: Disable / 1: Enable
0: Disable / 1: Enable
0: Disable / 1: Enable
0: Gross < -(Maximum capacity + 8d) /1: Gross < -19d
0: Disable / 1: Enable
0: 4-wire type / 1: 6-wire type

100

Checking software version

131415Checking display

16171999999
1000.001.0
211101

Digital calibration function list
Checking key switches

Executing zero calibration

Actual load calibration

Checking signal (mV/V)

Zero calibration

Executing span calibration

Digital calibration function

Span calibration

Calibration weightvalue input

Item selection

Setting value input

Power supply

DC24V -15% to +10%, 4.5W max.

Load cell input

Signal input range

DC5V ±5% 90 mA
Up to six 350 Ω load cells can be connected inparallel. 6-wire type with remote sensing-7.0 mV/V to +7.0 mV/V

minimum input sensitivity
Nonlinearity

0.15 μV/d or more (d=minimum division)
0.005% of F.S. max.

Temperature coefficient

Zero drift: ±0.02 μV/°C typ. ±0.1 μV/°C max.
Span drift: ±3 ppm/°C typ. ±15 ppm/°C max.

Sampling rate

1200 times / second

Excitation voltage

Display
Main display

7-digit LED (green) with a character height of 10 mm

Status display

LED (red) x 6Unit

Attach a label of g / kg / t

Setting value

Default

Key switches

01
02

Load cell input signal at Zero Calibration
Load cell input signal (at Span Calibration– at Zero Calibration)
Weight value at Span Calibration-7.00000 to 7.00000 mV/V
0.00001 to 7.00000 mV/V

0.00000
2.00000

External interface

1 to 999999

20000

Basic function list
Setting value input

Panel mount-10°C to +40°C
Less than 85%RH, non-condensing(When the indicator is installed to the control panel)
Front panel: IP65. Inside the panel: IP2X

Setting item

03

Item selection96(W) x 48(H) x 98.5(D) mm

Installation method
Operating temperature andhumidity range
IP ratingdCAL

Checking load cell input signal (mV/V)

Calibration function

Dimension

FncF

Setting item

01
02
03
04
05
06

Locking [ZERO/←] key
Locking [TARE/↑] key
Locking [F1/↓] key
Locking [F2/ ] key
Function of [F1/↓] key
Function of [F2/ ] key

07
08
09
10

Checking signal (mV/V)

Initialization mode
1112
13

Setting value

EtherNet/IP

AD-4411-PRT

PROFINETAD-4411-ECT

EtherCATUSB

Type-C connector, USB 2.0 (Full-speed)

Default

0: Disable / 1: Enable
0: Disable / 1: Enable
0: Disable / 1: Enable
0: Disable / 1: Enable
0: None / 1: Tare clear / 2: Zero clear /3: Gross / net display selection
4: High-resolution display selection
Condition to turn S1 status ON 0: None / 1: Hi / 2: OK / 3: Lo /Condition to turn S2 status ON 4: Zero setting error / 5: Taring error /Condition to turn S3 status ON 6: High resolution display
Digital filter cut-off frequency
0: 273.0
12: 20.0 24: 2.8
36: 0.40[Hz]
1: 120.0
13: 17.0 25: 2.4
37: 0.34
2: 100.0
14: 14.0 26: 2.0
38: 0.28
3: 84.0
15: 12.0 27: 1.7
39: 0.24
4: 70.0
16:10.0
28: 1.4
40: 0.20
5: 68.0
17: 8.4
29: 1.2
41: 0.17
6: 56.0
18: 7.0
30: 1.0
42: 0.14
7: 48.0
19: 6.8
31: 0.84 43: 0.12
8: 40.0
20: 5.6
32: 0.70 44: 0.10
9: 34.0
21: 4.8
33: 0.68 45: 0.08
10: 28.0
22: 4.0
34: 0.56 46: 0.07
11: 24.0
23: 3.4
35: 0.48000000Upper limit value
Lower limit value
Comparison target for Upperlimit value / Lower limit value

10-101-999999 to 999999-999999 to 999999
1: Gross / 2: Netx5AD-4411-EIP

00030

External dimension

96

991

44.6

Measurement mode

Setting value

89.5

92
+0.6-0Press more than 3s

Setting item

45

Power ONCALF

48

Operation mode

+0.8-0Panel thickness
1.6 to 3.2

Unit: mm

For functions other than those listed above, see the "AD-4411 Instruction Manual" onthe A&D website (https://www.aandd.jp).

FCC - Supplier's Declaration of Conformity
47 CFR § 2.1077 Compliance Information
Model: AD-4411
Responsible Party: A&D ENGINEERING, INC.
Address: 4622 Runway Boulevard Ann Arbor, MI 48108, U.S.A.
Tel: [1] (888) 726-5931
This device complies with part 15 of the FCC Rules. Operation is subject to thefollowing two conditions: (1) This device may not cause harmful interference, and(2) this device must accept any interference received, including interference thatmay cause undesired operation.