Wrist Digital Blood Pressure Monitor

Model UB-543
Instruction Manual
Original

Manuel d’instructions
Traduction

Manual de Instrucciones
Traducción

Manuale di Istruzioni
Traduzione

使用手冊
1WMPD4002838F

翻譯Contents

Dear Customers................................................................................................................................................... 2
Preliminary Remarks ........................................................................................................................................... 2
Precautions .......................................................................................................................................................... 2
Parts Identification ............................................................................................................................................... 4
Symbols ............................................................................................................................................................... 5
Using the Monitor................................................................................................................................................. 7
1. Installing / Changing the Batteries.............................................................................................................. 7
2. Selecting a User ......................................................................................................................................... 8
3. Adjusting the Built-in Clock Before Use...................................................................................................... 8
4. Applying the Cuff ........................................................................................................................................ 9
5. How to Take Accurate Measurements ....................................................................................................... 9
6. The C.P.G. Function Switch ...................................................................................................................... 10
7. Selecting the C.P.G. and my C.P.G. ........................................................................................................... 10
8. Measurement ........................................................................................................................................... 11
9. After Measurement ................................................................................................................................... 11
Measurements ................................................................................................................................................... 12
Recalling the Memory Data ............................................................................................................................... 13
Deleting Data Stored in Memory ........................................................................................................................ 14
What Is The IHB/AFib Indicator? ....................................................................................................................... 14
What Is The AFib? ............................................................................................................................................. 15
Pressure Bar Indicator ....................................................................................................................................... 15
WHO Classification Indicator ............................................................................................................................. 15
The C.P.G. Indicator .......................................................................................................................................... 16
About Blood Pressure ........................................................................................................................................ 17
What is Blood Pressure? .............................................................................................................................. 17
What is Hypertension and How is it Controlled? .......................................................................................... 17
Why Measure Blood Pressure at Home? ..................................................................................................... 17
WHO Blood Pressure Classification ............................................................................................................. 18
Blood Pressure Variations ............................................................................................................................ 18
Maintenance ...................................................................................................................................................... 18
Troubleshooting ................................................................................................................................................. 19
Technical Data ................................................................................................................................................... 20
English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure monitor, one of the most advanced monitors
available today. Designed for ease of use and accuracy, this device will facilitate your daily blood pressure regimen.
We recommend that you read through this manual carefully before using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical Products. This is made evident by themark of conformity. (0123: The reference number to the involved notified body)
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in the home healthcare environment.
This device is designed to measure blood pressure and pulse rate of people for diagnosis.

Precautions

Precision components are used in the construction of this device. Extremes in temperature, humidity, direct
sunlight, shock or dust should be avoided.
Clean the device with a dry, soft cloth or a cloth dampened with water and a neutral detergent. Never use alcohol,
benzene, thinner or other harsh chemicals to clean the device.
Avoid tightly folding the cuff for long periods, as such treatment may shorten the life of the components.
The device is not water resistant. Prevent rain, sweat and water from soiling the device.
Measurements may be distorted if the device is used close to televisions, microwave ovens, cellular telephones,
X-ray or other devices with strong electrical fields.
Wireless communication devices, such as home networking devices, mobile phones, cordless phones and their basestations, walkie-talkies can affect this blood pressure monitor.
Therefore, a minimum distance of 30 cm should be kept from such devices.
Used equipment, parts and batteries are not treated as ordinary household waste, and must be disposed ofaccording to the applicable local regulations.
When reusing the device, confirm that the device is clean.
Do not modify the device. It may cause accidents or damage to the device.
English 2

To measure blood pressure, the wrist must be squeezed by the cuff hard enough to temporarily stop blood flow
through the artery. This may cause pain, numbness or a temporary red mark to the wrist. This condition will appear
especially when measurement is repeated successively. Any pain, numbness, or red marks will disappear with time.
Measuring blood pressure too frequently may cause harm due to blood flow interference. Check that the operationof the device does not result in prolonged impairment of blood circulation, when using the device repeatedly.
If you have had a mastectomy, please consult a doctor before using the device.
Do not let children use the device by themselves and do not use the device in a place within the reach of infants. It
may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by mistake by infants.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially cause burns.
Allow the device to adapt to the surrounding environment before use (about one hour).
Clinical testing has not been conducted on newborn infants and pregnant woman. Do not use on newborn infants or
pregnant woman.
Do not touch the battery and the patient at the same time. That may result in electrical shock.
In the case of single components failure enclosure of near cuff may become hot and potentially cause malfunction.
Do not inflate without wrapping the cuff around your wrist.

Contraindications

The following are precautions for proper use of the device.
Do not apply the device to a wrist with other medical electrical equipment attached. The equipment may not
function properly.
People who have a severe circulatory deficit in the arm must consult a doctor before using the device to avoid
medical problems.
Do not self-diagnose the measurement results and start treatment by yourself. Always consult your doctor for evaluation of
the results and treatment.
Do not apply the device on a wrist with an unhealed wound.
Do not apply the device on an arm receiving an intravenous drip or blood transfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as a high-pressure oxygen chamber or an oxygen tent.
It may cause a fire or explosion.
English 3

Parts Identification
button ( average, memory and change of preset parameters)

Display

SET

button ( user selection

and clock

START

)

buttonBattery cover
Carrying case

Cuff

Alkaline batteries (LR03 or AAA)

Display
WHO classification indicatorandPressure bar indicatorHeart markUser

and User
MEMORY
Average

Battery indicator

Systolic pressure
Diastolic pressure
Correct Position Guidance (C.P.G.) indicator
IHB/AFib symbol
Pulse rate
Date & clock display
AM / PM markEnglish 4

Symbols
Symbols that are printed on the device caseSymbolsSN

IP

Function / MeaningStandby and Turn the device on
Battery installation guide
Direct current
Serial number
Date of manufacture
Type BF: Device and cuff are designed to provide special protection against electrical shocks.
EC directive medical device label
International protection symbol
WEEE label
Manufacturer
EU-representative
Refer to instruction manual / booklet
User selection
Keep drySymbols that appear on the display
Symbols

Function / Meaning / Recommended Action
The heart mark appears while measurement is in progress. It blinks when the pulse is detected.
Remain as still as possible.
IHB/AFib symbol appears when an irregular heartbeat is detected.
It may light when a very slight vibration like shivering or shaking is detected.
User 1 and user 2
English 5

Symbols
Symbols that appear on the display (continued)
Symbols

or

SYS
DIA
PUL
AM
PM

Function / Meaning
Previous measurements stored in MEMORY
Average data
FULL BATTERY
The battery power indicator during measurement
LOW BATTERY
The battery is low when it blinks
Unstable blood pressure due to movement during measurement

Recommended Action

Replace all batteries with new ones
when the indicator blinks.
Try the measurement again.
Remain very still during the measurement.

The systolic and diastolic values are within 10 mmHg of each other.
The pressure value did not increase during inflation.
Fasten the cuff correctly, and try the
The cuff is not applied correctly.
measurement again.
PUL. DISPLAY ERROR
The pulse is not detected correctly.
Remove the batteries and press the
START button, and then install the
Blood pressure monitor internal error
batteries again. If the error still
appears, contact the dealer.
Systolic blood pressure in mmHg
Diastolic blood pressure in mmHg
Pulse per minute
Data taken between 4:00 and 9:59
Data taken between 18:00 and 1:59
English 6

Using the Monitor

1. Installing / Changing the Batteries
1. Remove the battery cover.

2. Remove the used batteries and insert new batteries into thebattery compartment as shown, taking care that the polarities(+) and (-) are correct. Use only LR03 or AAA batteries.

Battery cover

3. Attach the battery cover.
Used batteries
New batteriesInsert the batteries as shown in the battery compartment.
If installed incorrectly, the device will not work.
When(LOW BATTERY mark) blinks on the display,replace all batteries with new ones. Do not mix old and newbatteries. It may shorten the battery life, or cause the device
to malfunction.
(LOW BATTERY mark) does not appear when thebatteries are drained.
The battery life varies with the ambient temperature and may be shorter at low temperatures. Generally, four new
LR6 batteries will last approximately for three months when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device are for testing monitor performance andmay have a limited life.
Remove the batteries if the device is not to be used for a long period of time. The batteries may leak and cause a
malfunction.
When removing the batteries, preset parameters (of clock, user and my C.P.G.) are reset.

English 7

2. Selecting a User
1. Press theUsing the Monitor

SET button when turning off the device. The indicatororTurn offis blinking.

SET button

SET button.
2. Select a user from user and user using thePress the START button to turn off the device.
After three minutes of non-operation, the device will turn off automatically.

Useror user
START button

3. Adjusting the Built-in Clock Before Use
1. Press and hold theSET button until the year starts blinking.

2. Select the year using the
button.
SETPress thebutton to set the current year and move to month/day selection.
The date can be set anywhere between the years 2010 and 2059.
3. Select the month using the
button.
SET button to set the current month and move to day selection.
Press thePress and hold
Year
Month

4. Select the day using the
button.
SETPress thebutton to set the current day and move to hour/minute selection.

Day

5. Select the hour using the
button.
SET button to set the current hour and move to minute selection.
Press theHour

6. Select the minute using the
button.
SETPress thebutton to proceed to "4. Applying the Cuff".

MinuteHolding down the
button will change the value continuously.
Pressing the START button will turn the device off anytime.

SET button

buttonSET button

buttonSET button

buttonbuttonSET button
SET button

buttonSET button

The operation proceeds to "6.
The C.P.G. Function Switch".

Notes: After three minutes of non-operation, the device will turn off automatically.
When the clock has not been set,is indicated for the clock display.
When removing the batteries, preset parameters (of clock, user and my C.P.G.) are reset.
English 8

4. Applying the Cuff

Using the Monitor

1. Wrap the cuff around your wrist about 1 cm above your hand as shown in the figure
at the right.

About 1 cm

2. Fasten the cuff tightly using the Velcro strip.
Note: For proper measurements, fasten the cuff tightly and measure on a bare wrist.

5. How to Take Accurate Measurements

For the most accurate blood pressure measurement:
Remain still and keep quiet during measurement.
Sit down in a comfortable position. Place your elbow on a table. Raise your hand so that the cuff is at the samelevel as your heart.
Relax for about five to ten minutes before measurement. If you are excited or depressed by emotional stress, themeasurement will reflect this stress as a higher (or lower) than normal blood pressure reading and the pulse
reading will usually be faster than normal.
Try to measure your blood pressure at about the same time every day.
An individual's blood pressure varies constantly, depending on what they are doing, what they have eaten and what
they drink can have a very strong and rapid effect on your blood pressure.
Do not measure immediately after physical exercise or a bath. Rest for twenty or thirty minutes before taking themeasurement.
Do not cross your legs. Keep your feet flat on the floor and straighten your back.
This device bases its measurements on the heartbeat. If you have a very weak or irregular heartbeat, the device
may have difficulty determining your blood pressure.
Should the device detect a condition that is abnormal, it will stop the measurement and display an error symbol.
See page 6 for the description of symbols.
This blood pressure monitor is intended for use by adults. Consult with your physician before using this device on achild. A child should not use this device unattended.
English 9

6. The C.P.G. Function Switch

Refer to page 16 for the C.P.G. function that will indicate the proper
angle so that the height of the cuff is the same level as your heart.

1. After step 6 in page 8, press the
button to select either " " or" " concerning the C.P.G. function.
2. Press the SET button to store the selection.
3.
If you do not use my C.P.G. function, press START button
to turn off. Proceed to "8. Measurement".
If you use my C.P.G. function, proceed to "7. Selecting the
C.P.G. and my C.P.G.".

The C.P.G. function
buttonPress the SET button
To "7. Selecting the C.P.G. and my C.P.G." or
START button and to "8. Measurement"

The C.P.G. indicator

7. Selecting the C.P.G. and my C.P.G.

You can select an indicator either the C.P.G. function or my C.P.G. .
Preset a proper posture (wrist angle) in memory if you use the my C.P.G. .

1. Adjust and keep the height of the blood pressure monitor to
the same level as your heart using your wrist angle.
2. Select an indicator using the
Indicator

button.

Step 1 [Adjust and keep the heightof the device (wrist angle) ]HIGH (Orange light)
Correct Height (Blue light).....The C.P.G. function is used. (my C.P.G. function : OFF )Data of my C.P.G. is deleted. Proceed to step 3.
Indicator

From step 6 of page 8.....my C.P.G. function is ON and the current angle is stored

when switched to the indicator . Proceed to step 3.
3. Press the START button to turn off the device.

English 10

LOW (Orange light)
Step 2 [Select The C.P.G. or my C.P.G.]
The C.P.G. function
My C.P.G.
buttonEx.: user 1
START button and to "8. Measurement"

Note:

When removing the batteries, preset parameters (of clock, user and my C.P.G.) are reset.
SET button.
Select a user from user and user using the

8. Measurement

During measurement, it is normal for the cuff to feel very tight.

9. After Measurement

While readings are displayed, if you press the START button to turn the device off, new readings are stored inmemory.
While readings are displayed, if you press the
button to turn the device off, new readings are not stored.
Remove the cuff and record your data.

Notes: The device is provided with an automatic power shut-off function which stores the current data in the memory
and turns the device off automatically one minute after measurement. Allow at least three minutes between
measurements on the same person.

English 11

Measurements
Note: The UB-543, once used, will provide an inflation appropriate to the user.
1. Wrap the cuff around your wrist. Sit comfortably with the cuff at the same levelas your heart and relax.
2. Press the START button. All of the display segments are displayed.
3. Select a user from user and user using the
Adjust and keep the height of the cuff (with blue light) to the same level as yourheart using the C.P.G. indicator.
Note: If you do not use user selection, wait for the inflation for several seconds.
If you do not use the C.P.G. function, the C.P.G. indicator is not displayed.

All segments

SET button immediately.

Useror user:

The C.P.G. indicator: Orange Blue Orange
LOW LEVEL HIGH

4. Zero (0) is displayed blinking briefly. Then the display changes, as
At heart levelmeasurement begins. The cuff starts to inflate. It is normal for the cuff
to feel very tight. The measurement starts automatically when
Zero display
inflation starts, and the(heart mark) blinks.
Inflation is started
Note: If you wish to stop inflation at any time, press the START button again.
5. When the measurement is complete, the device displays readings( of the systolic and diastolic pressure readings, pulse rate, WHOclassification, IHB/AFib symbol and C.P.G. indicator). While readings
are displayed, the date and time are displayed alternately. The cuff
exhausts the remaining air and completely deflates automatically.
Note: If you do not want to store new readings in memory, press the
button while readings are displayed.
6. Press the START button again to turn the device off. Remove the cuff.
Notes: The device is provided with an automatic power shut-off function.
Allow at least one minute between measurements on the same person.
English 12

SET button

Inflating and
Measurement in progress
Systolic pressure
Diastolic pressure
WHO classification
IHB/AFib symbol
Pulse rate
The C.P.G. indicatorRecalling the Memory Data

Turn off

Note: This device stores the last sixty sets of data in memory automatically.
1. Press the
button when turning the device off.
The average of all measurements and the number of data are displayed.
If no data, “0” is displayed. Press theor START button to turn thedevice off.
2. Use the following buttons to display data (of number and measurement data).
SET button.
Select a user from user and user using the
The device displays the average of all measurements and thenumber of data are displayed.
Pressing the
button each time , the device displays as follows:
Average data of all AM (morning) measurements taken between
4:00 and 9:59. In the example, If no data,
is displayed.
Average data of all PM (evening) measurements taken between
18:00 and 1:59.
Data (of number and measurement data).
The device displays in order from most recent data. The date
and time are displayed alternately while displaying themeasurement data.
In the example: No.30 & data
No.29 & data
No.01 & data.
3. If you press the
button after oldest data is displayed, the device
proceeds to step 1, the average of all measurements and the number ofdata are displayed.
4. Press the START button to turn the device off.
After one minute of non-operation, the device will turn off automatically.
English 13

Memory mode
User selection
Example: user

buttonbuttonAverage of all the data
Number of all data
Average for AM data
AM indicator
Number of AM data
Average for PM data
PM indicator
Number of PM data
Ex.: No.30
Most recent data
Date
Time
3 seconds
Last data (Oldest)
Date
Time
3 seconds

buttonbuttonbuttonbuttonbuttonbuttonTurn offDeleting Data Stored in Memory
1. Select a user from user and user
device using the START button.

using theSET button. Turn off theTurn off
SET button

2. Press and hold the
button until truing off the device automatically.
The device displays a user icon and themark, deletes data stored inmemory while blinking themark and turns off automatically.

User

Note: This operation will delete specified user data stored in memory.
You cannot select which data to delete.

Display is turned off

or userPress and hold button
Ex.: User
Deleting data
Display is turned off
Release button

What Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements, the IHB/AFib indicator will appear on thedisplay with the measurement values.
Note: We recommend contacting your physician if you see thisEnglish 14

IHB/AFib indicator frequently.

What Is The AFib?
The heart contracts due to electrical signals occurring in heart and sends blood through the body. Arterial fibrillation(AFib) occurs when the electrical signal in the atrium becomes confused and leads to disturbances in the pulseinterval. AFib can cause blood to stagnate in the heart, which can easily create clots of blood, a cause of stroke andheart attack.
Pressurizing

Pressure Bar Indicator
The indicator monitors the progress of pressure during
measurement.

WHO Classification Indicator
Each six segments of the bar indicator correspond to the
WHO blood pressure classification described on 18 page.
ExampleModerate
hypertension

Mild hypertension

High normal

Inflation in progress
Measurement in progress
Inflation complete
WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimal: The indicator displays a segment, based on thecurrent data, corresponding to the WHO
classification.

English 15

The C.P.G. Indicator

The C.P.G. Indicator
The C.P.G. indicator

The C.P.G. (Correct Position Guidance) indicator is the function to inform adifference between the height (wrist angle) of the blood pressure monitor andyour cardiac height in the correct posture (Example: sitting posture, height oftable and chair, etc.) during the measurement. The indicator can be used toget more stable measurement condition.

The C.P.G. Indicator
The height of the blood pressure
monitor is lower than your heart.
Low angle

Incorrect Height

The C.P.G. indicator : LOW

The height of the blood pressure
monitor is the same level as yourheart.
Proper angle
Correct Height

The height of the blood pressure
monitor is higher than your heart.

The C.P.G. indicator : LEVELThe C.P.G. indicator : HIGH

The position of the device is checked both before and after measurement.
If both checks show a correct measurement position the LEVEL indicator islit(blue)
For all other measurements an indicator for LOW or HIGH measurement
position will be lit (orange)

How To Use My C.P.G.

The C.P.G. function can be used with proper posture (wrist angle) in the
majority of measurements. If you need to change the posture to adjust theheight so that the height of the blood pressure monitor is the same level as
your cardiac height, you can use my C.P.G. function to store a personal
posture. Preset your angle to my C.P.G. function before measurement.
English 16

High angle

Incorrect Height

Proper Posture
Sit straight back on achair and put yourelbow on a table.

Table

Chair

The Indicator During Measurement And Recalling Memory

The C.P.G. indicator can be displayed at the beginning of measurement and is included
in data stored in the memory. Adjust and keep the angle during measurement.

What is Blood Pressure?

About Blood Pressure

Blood pressure is the force exerted by blood against the walls of the arteries. Systolic pressure occurs when theheart contracts. Diastolic pressure occurs when the heart expands. Blood pressure is measured in millimeters of
mercury (mmHg). One's natural blood pressure is represented by the fundamental pressure, which is measured
first thing in the morning while one is still at rest and before eating.

What is Hypertension and How is it Controlled?

Hypertension, an abnormally high arterial blood pressure, if left unattended can cause many health problems
including stroke and heart attack. Hypertension can be controlled by altering lifestyle, avoiding stress, and withmedication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?

Blood pressure measured at a clinic or doctor's office may cause apprehension and can produce an elevatedreading, 25 to 30 mmHg higher than that measured at home. Home measurement reduces the effects of outside
influences on blood pressure readings, supplements the doctor's readings and provides a more accurate, complete
blood pressure history.

English 17

WHO Blood Pressure Classification

Standards to assess high blood pressure, without regard to age,have been established by the World Health Organization(WHO), as shown in the chart at the right.

Blood Pressure Variations

An individual’s blood pressure varies greatly on a daily and seasonalbasis. It may vary by 30 to 50 mmHg due to various conditions
during the day. In hypertensive individuals variations are even more
pronounced. Normally, the blood pressure rises while at work or playand falls to its lowest levels during sleep. So, do not be overly
concerned by the results of one measurement.
Take measurements at the same time every day using theprocedure described in this manual to get to know your normal bloodpressure. Regular readings give a more comprehensive bloodpressure history. Be sure to note the date and time when recording
your blood pressure. Consult your doctor to interpret your bloodpressure data.

MaintenanceDo not open the device. It uses delicate electrical components and an intricate air unit that could be damaged. If you
cannot fix the problem using the troubleshooting instructions, contact the authorized dealer in your area or ourcustomer service department. The A&D customer service will provide technical information, spare parts and units toauthorized dealers.
The device was designed and manufactured for a long service life. However it is generally recommended to havethe device inspected every 2 years, to ensure proper functioning and accuracy. Please contact the authorized
dealer in your area or A&D for maintenance.
English 18

Troubleshooting
ProblemPossible Reason
Batteries are drained.

Nothing appears in thedisplay, even when the Battery terminals are not in the correct
device is turned on.
position.
The cuff does not
inflate.

Battery voltage is too low.
(LOW BATTERY mark) blinks. If
the batteries are drained completely, themark does not appear.
The cuff is not applied properly.
You moved your wrist or body during themeasurement.

Recommended Action
Replace all batteries with new ones.
Reinstall the batteries with negative and positive
terminals matching those indicated in the battery
compartment.
Replace all batteries with new ones.

Apply the cuff correctly.
Make sure you remain very still and quiet during
the measurement.
The device does not
Sit comfortably and still.
measure.
Place your elbow on a table with your palm facing
Readings are too high The cuff position is not correct.
upward and the cuff is at the same level as your heart.
or too low.
If you have a very weak or irregular heartbeat, thedevice may have difficulty in determining yourblood pressure.
The value is different from that measured Refer to “Why Measure Blood Pressure at
at a clinic or doctor’s office.
Home?”.
Other
Remove the batteries. Place them back properly
and try the measurement again.
Note: If the actions described above do not solve the problem, contact the dealer. Do not attempt to open or repair
this product yourself, as any attempt to do so will make your warranty invalid.
English 19

Technical DataType
Measurement method
Measurement range
Measurement accuracy
Power supply
Number of measurements
Wrist circumference
Classification
Applied part
Useful life
Clinical test
EMD
MemoryOperating conditions
Transport / storage conditions
Dimensions
WeightIngress protection

UB-543
Oscillometric measurement
Pressure: 0 – 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 – 180 beats / minutePressure: ±3 mmHg
Pulse:±5 %2 x 1.5 V alkaline batteries (LR03 or AAA)Approx. 180 measurements, when AAA alkaline batteries are used, with pressure valueof 170 mmHg at room temperature of 23 °C.
13.5 – 21.5 cm
Internally powered ME equipment (Continuous operation mode)
Cuff Type BF
Device: 5 years (when used six times a day)
According to ISO81060-2：2013
IEC 60601-1-2: 2014
Last 60 measurements each for user 1 and user 2.
+10 to +40 °C / 15 to 85 %RH / 800 to 1060 hPa-20 to +60 °C / 10 to 95 %RH / 700 to 1060 hPa
Approx. 56 [W] x 88 [H] x 18 [D] mm
Approx. 104 g, excluding the batteries
IP20

Note: Specifications are subject to change for improvement without prior notice.
IP classification is the degrees of protection provided by enclosures in accordance with IEC 60529. This device
is protected against solid foreign objects of 12 mm diameter and greater such as a finger. This device is notprotected against water.
English 20

A&D Company, Ltd.
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119
Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
Telephone: [31] (70) 345-8570 Fax: [31] (70) 346-7299
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119
A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409
OOO A&D RUS
OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66
A&D Technology Trading(Shanghai) Co. Ltd

爱安德技研贸易（上海）有限公司

中国 上海市自由贸易试验区浦东南路855号世界广场32楼C,D室 邮编200120(32CD, World Plaza, No.855 South Pudong Road,China (Shanghai) Pilot Free Trade Zone, 200120, China)
电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599