Wrist Digital Blood Pressure Monitor
Model UB-1100BLE
Instruction Manual English
Original

Manuel d’instructions Français
Traduction

Manual de Instrucciones Español
Traducción
Manuale di Istruzioni
Traduzione

BedienungsanleitungÜbersetzung

使用手冊
翻譯

‫دليل اإلرشادات‬‫الترجمة‬1WMPD4003739D

Italiano
Deutsch中文‫العربية‬Contents

Dear Customers................................................................................................................................................... 2
Preliminary Remarks ........................................................................................................................................... 2
Precautions .......................................................................................................................................................... 2
Parts Identification ............................................................................................................................................... 5Contents .............................................................................................................................................................. 5
Symbols ............................................................................................................................................................... 6
Using the Monitor............................................................................................................................................... 10
Measurements ................................................................................................................................................... 17
What Is an IHB Indicator? .................................................................................................................................. 21
What Is an Irregular Heartbeat .......................................................................................................................... 21
WHO Classification Indicator ............................................................................................................................. 21
About Blood Pressure ........................................................................................................................................ 22
Settings by App.................................................................................................................................................. 24
Settings .............................................................................................................................................................. 25
Troubleshooting ................................................................................................................................................. 27
Maintenance ...................................................................................................................................................... 28
Technical Data ................................................................................................................................................... 28

English 1

Dear Customers

Congratulations on purchasing a state-of-the-art A&D blood pressure monitor.Designed for ease of use andaccuracy, this monitor will facilitate your daily blood pressure regimen.
We recommend that you read through this manual carefully before using the device for the first time.
The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any use of suchmarks by A&D is under license. Other trademarks and trade names are those of their respective owners.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical Products. This is mademark of conformity. (0123: The reference number to the involved notified body)evident by the

Hereby, A&D Company, Limited declares that the radio equipment type UB-1100BLE is in compliance with
Directive 2014/53/EU. The full text of the EU declaration is available at the following internet address:
http://www.aandd.jp/products/manual/manual_medical.html
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in the home healthcare environment.
This device is designed to measure blood pressure and pulse rate of people for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremes in temperature, humidity, direct
sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened with water and a neutral detergent. Never use
alcohol, benzene, thinner or other harsh chemicals to clean the device or cuff.
English 2

The device and cuff are not water resistant. Prevent rain, sweat and water from soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions, microwave ovens, cellular telephones,
X-ray or other devices with strong electrical fields.
When reusing the device, confirm that it is clean.
Used equipment, parts and batteries are not treated as ordinary household waste, and must be disposed ofaccording to the applicable local regulations.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the wrist must be squeezed by the cuff hard enough to temporarily stop blood flow
through the artery. This may cause pain, numbness or a temporary red mark to the wrist. This condition willappear especially when measurement is repeated successively.
Any pain, numbness, or red marks will disappear with time.
If you feel strong pain during the measurement, please remove the cuff.
Measuring blood pressure too frequently may cause harm due to blood flow interference.
Avoid tightly folding the cuff for long periods, as such treatment may shorten the life of the components.
When the AC adapter is used, make sure that it can be readily removed from the electrical outlet whennecessary.
Sometimes the temperature of the cuff can reach 44°C in an ambient temperature of 40°C. If you feel the cuff is
too hot, please remove the cuff immediately.
Please do not simultaneous use HF surgical equipment.
Clinical testing has not been conducted on newborn infants and pregnant woman. Do not use on newborn infants
or pregnant woman.
If you have had a mastectomy, please consult a doctor before using the device.
Do not let children use the device by themselves and do not use the device in a place within the reach of infants. It
may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by mistake by infants.
Do not touch the batteries and the patient at the same time. That may result in electrical shock.
Wireless communication devices, such as home networking devices, mobile phones, cordless phones and theirbase stations, walkie-talkies can affect this blood pressure monitor. Therefore, a minimum distance of 30 cm
English 3

should be kept from such devices.
Unplug the ac adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially cause bums.
Allow the device to adapt to the surrounding environment before use (about one hour).
Do not inflate without wrapping the cuff around the Wrist.
The automatic blood pressure monitor's performance may be affected by excessive temperature or humidity, or
altitude.

Contraindications

The following are precautions for proper use of the device.
Do not apply the cuff on wrist in which another medical device is attached. The equipment may not function
properly.
People who have a severe circulatory deficit in the wrist must consult a doctor before using the device, to avoid
medical problems.
Do not self-diagnose the measurement results and start treatment by yourself. Always consult your doctor for evaluation of
the results and treatment.
Do not apply the cuff on wrist with an unhealed wound.
Do not apply the cuff on wrist receiving an intravenous drip or blood transfusion. It may cause injury.
Do not use the device where flammable gases such as anesthetic gases are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as a high-pressure oxygen chamber or an oxygen
tent. It may cause a fire or explosion.

English 4

Parts Identification
RED LED
Start/Stop button

OLED
Reset button

Cuff

USB Connector

Contents
・1 Blood Pressure Monitor
・1 Instruction Manual
・1 Charging Cable
・1 QR code card
English 5

Symbols
Symbols that are printed on the device case and cuff
Symbols

Function / MeaningStandby and Turn the device on.
Direct current
Type BF: Device and cuff are designed to provide special protection against electrical shocks.

IP

SN
BT

EC directive medical device label
EU-representative
Manufacturer
Date of manufactureInternational protection symbol: IP classification is degrees of protection provided by enclosures inaccordance with IEC 60529. This device is protected against solid foreign objects of 12mm diameter andgreater such as a finger. This device is not protected against water.
WEEE label
Serial number
Bluetooth address
Refer to instruction manual/booklet
To indicate generally elevated, potentially hazardous, levels of non-ionizing radiation, or to indicate
equipment or systems e.g. in the medical electrical area that include RF transmitters or that intentionally
apply RF electromagnetic energy for diagnosis or treatment.
Keep dry
Universal Serial Bus

English 6

Symbols that appear on the display
Symbols
SYS
DIA
PUL

Function / Meaning
Systolic blood pressure in mmHg

Recommended Action

Diastolic blood pressure in mmHg
Pulse per minute
The battery power is low

Charge the device

User Select mode

Select the user

Application is connected with the device
Application has disconnected from the device
C.P.G(Correct Position Guidance)
The C.P.G. Indicator:
LOW. The height of the blood pressure monitor
Is lower than your heart.
The C.P.G. Indicator:
LEVEL. The height of the blood pressure monitor
Is the same level as your heart.
The C.P.G. Indicator:
HIGH. The height of the blood pressure monitor ishigher than your heart.

E PUL
/min

Place the blood pressure monitor at the samelevel as your heart.

Place the blood pressure monitor at the same levelas your heart.

Appears while measurement is in progress.

Measurement is in progress. Remain as still as
possible.

The pulse is not detected correctly.

Apply the cuff correctly, and take anothermeasurement.
English 7

Symbols that appear on the display (continued)
Symbols
Err 1
Err 3
Err 9

Function / Meaning
Unstable blood pressure due to movement during
measurement. The systolic and diastolic values arewithin 10mmHg of each other.
The pressure value did not increase during inflation.
The cuff is not applied correctly.
Blood pressure monitor internal error.

Recommended Action
Take another measurement.
Remain still during measurement.
Apply the cuff correctly, and take anothermeasurement.
Press the reset button.
After reset, clock data will disappear.
If the error still appears, contact A&D Medical.

Data is transferring
Measurements can be taken but device will nottransmit data wirelessly.
The reading may yield an incorrect value.
Appears when a body or wrist movement is detected. Take another measurement.
Remain still during measurement.
The reading may yield an incorrect value.
Appears during measurement when the cuff isApply the cuff correctly, and take anotherattached loosely.
measurement.
Airplane mode

Firmware is updating.
Appears when irregular heart beat is detected.
Appears when your arm was in the Incorrect position
Take another measurement and ensure your wrist is atduring measurement, either higher than your heart or
the same height as your heart.
lower than your heart.
English 8

Symbols that appear on the display (continued)
Symbols

Function / Meaning

Recommended Action

Appears when your arm is in thecorrect position during measurement.

English 9

Using the Monitor

UB-1100BLE Digital Blood Pressure Monitor features 5 measurement modes.
1.

In-App Mode Measurement:
User can take measurements and view their trends in the A&D Connect app.

2.

Offline Mode Measurement:
Measurement can also be completed without the A&D Connect app.
Data will be stored on the monitor and transferred to the app on next connection.

3.

Multiuser Mode Measurement(In-app):Up to 6 users (5 accounts and 1 visitor) can use and store data on the app. Each user must have a mobile device
paired with the monitor.

4.

Guest Mode Measurement(In-app):User can take a measurement as Guest, without account registration in the A&D Connect app. The data is savedon the mobile device memory but not in your account.

5.

Visitor Mode Measurement (Offline):
Select “0 Visitor” to take a measurement in visitor mode from the monitor only. Data is not stored in visitor mode.

English 10

Charging the Device

Using the Monitor

Battery may be partially charged but be sure to charge the device completely upon first use after purchase.
1. Remove the USB connector cover.

2. Connect the charging cable.

3. During charging, the red LED will illuminate.

4. When charging is complete, the red LED will turn off.

5. Once charging is complete, remove the charging cable.

English 11

Using the Monitor
CAUTION!
 The battery life varies with the ambient temperature and may be shorter at low temperatures.
 Please use the specified charging cable shipped with the device.
 Blood pressure measurement cannot be done during charging.
 When disconnecting the AC adapter from the electrical outlet, grasp and pull the AC adapter body out
the outlet.
 When disconnecting the charging cable from the blood pressure monitor, grasp and pull the charging
cable out of outlet.
 Do not connect to equipment other than those specified.
 Do not change the battery.
 If the battery can no longer charged, please contact the dealer.
 Overcharging the battery may reduce its lifetime.
 Lithium battery replacement by inadequately trained personnel could result in a hazard such as a fireor explosion.

English 12

Wireless Function
In the unlikely event that this monitor causes radio wave interference to a different wireless station, change the location
where this monitor is used or stop use immediately.

Be sure to use in a location where visibility between the two devices that you want to connect is good. The connection
distance is reduced by the structure of buildings or other obstructions. In particular, connection may be impossible
when devices are used on either side of reinforced concrete.



Do not use Bluetooth® connection in the range of a wireless LAN or other wireless devices, near devices that emit
radio waves such as microwaves, in locations where there are many obstructions, or in other locations where signal
strength is weak. Doing so may result in frequent loss of connection, very slow communication speeds and errors.



Using close to an IEEE802.11g/b/n wireless LAN device may cause mutual interference to occur, which may result in
reduced communication speeds or which may prevent connection.
In this case, switch off the power supply to the device that is not being used, or use the monitor in a different location.



If the monitor does not connect normally when used near a wireless station or broadcast station, use the monitor in adifferent location.



A&D Company, Limited cannot accept liability for any damages incurred due to impaired operation or data loss, etc. that
occur through the use of this product.

This product is not guaranteed to connect to all Bluetooth® compatible devices.

English 13

To connect with your mobile device – download and install the “A&D application” onto your mobile device.
Later referred to as “APP”.

For the latest compatibility list visit

https://www.aandd.co.jp/products/hhc/hhc-ict_compatibility.html

English 14

Device Setup (Pairing)

Using the Monitor

1. Ensure the Bluetooth® function of the mobile device is ON.
2. Launch up A&D Medical Connect app.
The icon to the left will
Blink on the monitor.
3. Accept the paining request from the pop-up window.

Multiuser Pairing
The monitor can be registered for up to 5 users. Follow the above instructions to pair the mobile device for each user.
New users can be added anytime. The user who has the lowest frequency of measurement times will be cleared
once the device has reached the 5 user max.
Note: If new user uses the same mobile device, then re-paining is not required as this has been complete.
Note: Default is in visitor non memory mode with out paining.

CAUTION!
The communication distance between this monitor and the receiver device is about 33 feet (10m). Thisdistance can be reduced by the conditions in the surrounding environment, so be sure to check that the
distance is short enough for a connection to be made after measurement is complete.
English 15

Applying the Wrist Cuff

Using the Monitor

1. Wrap the cuff around your wrist about 1cm above your hand as shown in the figure below.
2. Apply the cuff tightly using the Velcro strip.
Note: For accurate measurements, apply the cuff tightly and measure on a bare wrist.

How to Take Accurate Measurements

For the accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross your legs. Keep your feet on the floor andstraighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same height as your heart.
Remain still and keep quiet during measurement.
Do not measure immediately after physical exercise or a bath. Rest for twenty or thirty minutes before taking themeasurement.
Try to measure your blood pressure at the same time every day.
English 16

In-App Mode Measurement

Measurements

1.

Ensure the Bluetooth® function of the mobile device is ON.

2.

Open the A&D Connect app. The app will automatically detect thepresence of the monitor.

3.

Press the Take Measurement button, followed by the Start buttonon the app screen.
User number

Username
Take measurement

App Display

Start

Monitor Display

4.

The C.P.G. indicator will display the correct wrist position briefly.

5.

The cuff starts to inflate. It is normal for the cuff to feel tight. The app displays the pressure value once inflation starts.

6.

When the measurement is complete, the app and the monitor will display
results similar to the icon on the right.
(Systolic, diastolic and pulse rate will be displayed in rotation.)

Note:

Press the start button again to turn the monitor off. The monitor is also
provided with an automatic power shut-off function. Allow at least
one minute between measurements on the same person.

English 17

Monitor Display

Result Notification
Icons(Refer to pg.8,21
For details)

App Display

Measurements
Offline Mode Measurement
1.

Press the Start/Stop button on the monitor.
After the A&D logo appears for 1 second, the User Select screen will display.

2.

Press the Start/Stop button again to select a user or switch between username from User Select screen.
Username can be changed in the app under Settings.
When this is not set, the first name will display as the username.

User number

Select User

3.

Measurement starts automatically after 3 seconds.

4.

Zero(0) is displayed blinking briefly. Then the display changes, as measurement begins.

5.

The cuff starts to inflate. It is normal for the cuff to feel tight. Monitor displays thepressure value once inflation starts.

6. When the measurement is complete, the monitor will display result similar to the icon on
the right. (Systolic, diastolic and pulse rate will scroll.)
Note: Press the start/stop button again to turn the monitor off. The monitor is also provided with
an automatic power shut-off function. Allow at least one minute between measurementson the same person.

English 18

User nameMeasurements
Upload Stored Measurement
The monitor can store up to 100 readings per user and can be transferred tothe app on next connection.
To transfer data, open the A&D Connect app and select Upload Measurementfrom the dashboard pop-up window.

Upload Measurement

App Display

Multiuser Mede Measurement
1. Up to 5 people can use and store readings on the app. Each user must have a mobile device paired with themonitor. (See pg.15 for pairing instructions.)
2. If you are taking the reading without the app, press the Start/Stop button on themonitor to switch between usernames. Select the desired user before
each reading. Data will transfer to the app on your next connection.
Note: Usernames will be displayed in order of greater usage frequency.

English 19

Measurements
Visitor Mode Measurement
1.
2.
3.
4.

Press the Start/Stop button on the monitor.
Press the Start/Stop button again to switch between username from User Select “0 Visitor”.
The C.P.G. indicator will display the correct wrist position briefly. Measurement starts automatically after 3 seconds.
When the measurement is complete, the monitor will display result similar to the icon on the right.
(Systolic, diastolic and pulse rate will be displayed in rotation.)
Visitor Mode is only available through the monitor. Data is not stored in visitor mode.

Notes for Accurate Measurement

Sit down in a comfortable position. Place your arm on a table with your palm facing upward and the cuff at thesame level as your heart.
Relax for about five or ten minutes before taking a measurement. If you are excited or depressed by emotional
stress, the measurement will reflect this stress as a higher (or lower) than normal blood pressure reading and thepulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you are doing and what you have eaten.
What you drink can have a very strong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very weak or irregular heartbeat, the device
may have difficulty determining your blood pressure.
Should the device detect a condition that is abnormal, it will stop the measurement and display an error symbol.
See page 7,8 for the description of the symbols.
This blood pressure monitor is intended for use by adults. Consult with your physician before using this device ona child. A child should not use this device unattended.
The automatic blood pressure monitor’s performance may be affected by excessive temperature or humidity, or
altitude.
English 20

What Is an IHB Indicator?

When the monitor detects an irregular rhythm during the measurements, the IHB indicator will appear on the display with themeasurement values.
Note: We recommend contacting your physician if you see thisIHB indicator frequently.

What Is an Irregular Heartbeat

The UB-1100BLE blood pressure monitor provides a blood pressure and pulse rate measurement even when an
irregular heartbeat occurs. An irregular heartbeat is defined as a heartbeat that varies from the average of allheartbeats during the blood pressure measurement.

WHO Classification Indicator
This Indicator’s by App:
Each segment of the bar Indicator corresponds to the
WHO blood pressure classification described on thenext page.

English 21

About Blood Pressure
What is Blood Pressure?

Blood pressure is the force exerted by blood against the walls of the arteries. Systolic pressure occurs when the heart
contracts. Diastolic pressure occurs when the heart expands. Blood pressure is measured in millimeters of mercury(mmHg). One's natural blood pressure is represented by the fundamental pressure, which is measured first thing in themorning while one is still at rest and before eating.

What is Hypertension and How is it Controlled?

Hypertension, an abnormally high arterial blood pressure, if left unattended, can cause many health problems including
stroke and heart attack. Hypertension can be controlled by altering lifestyle, avoiding stress, and with medication under adoctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?

Blood pressure measured at a clinic or doctor's office may cause apprehension and can produce an elevated reading, 25
to 30 mmHg higher than that measured at home. Home measurement reduces the effects of outside influences on bloodpressure readings, supplements the doctor's readings and provides a more accurate, complete blood pressure history.

English 22

WHO Blood Pressure Classification

Standards to assess high blood pressure, without regard to age,have been established by the World Health Organization (WHO),as shown in the chart.

Blood Pressure Variations

An individual’s blood pressure varies greatly on a daily and seasonal basis. It may vary by 30 to 50 mmHg due to various
conditions during the day. In hypertensive individuals’ variations are even more pronounced. Normally, the bloodpressure rises while at work or play and falls to its lowest levels during sleep. So, do not be overly concerned by the
results of one measurement.
Take measurements at the same time every day using the proceduredescribed in this manual to get to know your normal blood pressure.
Regular readings give a more comprehensive blood pressure history. Be
sure to note date and time when recording your blood pressure. Consult
your doctor to interpret your blood pressure data.

English 23

Settings by App
Setting
Nickname

Start Time

Remove User
Check Software
UpdateUnpairDescription
User can overwrite the default monitor
display username and assign a personal
username (up to 10 characters).
Enables a user to choose how long of an interval
between pressing “Start/Stop” button and when the monitor will start measurement. For
multiple user, a longer interval is useful. For single users, a shorter interval is useful. Default for
single user is set at 1 second and for multiple user is set at 5 seconds.
Allows user to remove himself as a user in the BP monitor.
Enables users to update the monitor with the most recent Bluetooth® software.

C.P.G.

Allows user to unpair his mobile device with the BP monitor. This can also be
accomplished via theBluetooth® setting in the mobile device.
This function will reset the monitor to factory settings. All date in the monitor will be deleted.
This will not affect any data stored in your mobile device.
Allows user to Enable/Disable the Correct Position Guidance indicator.

My Position

Allows user to set position for correct wrist angle

Reset

English 24

Settings

The C.P.G. Indicator Setting

The C.P.G. (Correct Position Guidance) indicator is the function to inform a difference
between the height (wrist angle) of the blood pressure monitor and your cardiac height in thecorrect posture (Example: sitting posture, height of table and chair, etc.) during themeasurement. The indicator can be used to get a more stable measurement condition.
The C.P.G. Indicator

LEVEL
LOW

The height of the blood pressure
monitor is at the same level as
your heart.

The height of the bloodpressure monitor is lower
than your heart.

Low angle

HIGH

Incorrect
HeightProper angle

Correct
Height

The height of the bloodpressure monitor is higher
than your heart.
High angle

Incorrect
HeightSettings

The position of the device is checked both before and after measurement. The correct
Measurement position is indicated by “arm mark logo” on the A&D Connect app.

My Position
Your C.P.G. allows you to set position for correct wrist angle for each user.
Press the setting icon in the Take Measurement screen of the wrist monitor.
Select My Position and then Set my Position.

App Display

English 25

Settings
Press Next to set your C.P.G. position.
Have the wrist monitor at your heart level
And press Set.
The blood pressure monitor now memorizes your best position.
You’ll see a check mark in the Set my Position box indicating it’s done.

Press Next
to set position

Monitor Setting

App Display

The airplane mode feature can be used in an environment than does not allow the emittance of radio frequency.
Measurement can be done in airplane mode and data will transfer to the app on next connection.
1. Hold the Start/Stop button and the airplane mode select screen will display.
Tap the Start/Stop button to switch between ON and OFF.
2. Select ON and wait 3 seconds to enter airplane mode.
In airplane mode, the airplane icon to the left will be illuminated.

English 26

Troubleshooting
SymptomPowerdoes notturn on.

Probable Cause
Low battery
Blood Pressure monitor
internal error.
Battery charging cable isconnected.

Treatment
Charge the battery.
Press the Reset button.
Remove the charging cable.

The cuff
Low Battery
Charge the battery.
does notinflate.
The device You moved your wrist or body Make sure you remain still and quiet during measurement.
during measurement.
does notIf you have a very weak or irregular heartbeat, the device may have difficult
measure.
in determining your blood pressure.
Readings
Sit comfortably and still.
are too
The cuff position is not
Place your arm on a table with your palm facing upward and the wrist cuff athigh or too correct.
the same level as your heart.
low.
At a clinic or doctor’s office apprehension may cause an elevated reading.
The value is different from
Home measurement reduces the effects of outside influences on bloodthat measured at a clinic orpressure readings, supplementing the doctor’s readings.
doctor’s office.
Note: If the actions described above do not solve the problem, contact the dealer. Do not attempt to open or repair
this product, as any attempt to do so will make your warranty invalid.
Other

English 27

MaintenanceDo not open the device. It uses delicate electrical components and an intricate air unit that could be damaged. If
you cannot fix the problem using the troubleshooting instructions, contact the authorized dealer in your area or ourcustomer service department. The A&D customer service will provide technical information, spare parts and units
to authorized dealers.
The device was designed and manufactured for a long service life. However it is generally recommended to havethe device inspected every 2 years, to ensure proper functioning and accuracy. Please contact the authorized
dealer in your area or A&D for maintenance.

Technical DataType
Measurement method
Measurement range

Measurement accuracy
Wrist circumferencePower supply
Number of measurements
Classification
Clinical test
EMD

UB-1100BLE
Oscillometric measurement
Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse: 40 - 180 beats / min
Pressure: ±3 mmHg Pulse: ±5 %13.5 cm - 21.5 cm
3.7 V Li-ion 325 mAh
Approx. 100 times (With pressure value 170 mmHg, room temperature 23 °C)
Internally powered ME equipment (Supplied by batteries)
Class II (Supplied by adapter)
Continuous Operation mode
According to ISO81060-2
IEC 60601-1-2: 2014
English 28

Technical Data
Wireless communication

Operating conditions
Transport / Storage conditions
Dimensions
WeightIngress protection
Applied part
Useful life
Included items

OptionZY (MURATA Manufacturing Co. Ltd.)
Bluetooth® Ver.4.1 Low Energy
Technology BLP
Frequency band: 2402 MHz to 2480 MHz
Maximum RF output power: 2.1 dBm
Modulation:GFSK+10 °C to +40 °C / 15 %RH to 85 %RH/800 hPa to 1060 hPa-20 °C to +60 °C / 10 %RH to 95 %RH/700 hPa to 1060 hPa
Approx. 54 [W] x 76 [H] x 21 [D] mm
Approx. 92 g
Device: IP20
Cuff Type BF
5 years (when used six times a day)UB -1100BLE - 1PC
Instruction Manual - 1PC
Charging cable(KSUSB-2C) - 1PC
QR code card-1PC
AC adapter: AX-TB283

English 29

MEMO

English 30

A&D Company, Ltd.
1-243 Asahi, Kitamoto-shi, Saitama-ken, 364-8585, JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119
Emergo Europe B.V.
Westervoortsedijk 60, 6827 AT Arnhem, The Netherlands
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119
A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409
OOO A&D RUS
OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66
A&D Technology Trading(Shanghai) Co. Ltd

爱安德技研贸易（上海）有限公司

中国 上海市自由贸易试验区浦东南路855号世界广场32楼C,D室 邮编200120(32CD, World Plaza, No.855 South Pudong Road,China (Shanghai) Pilot Free Trade Zone, 200120, China)
电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599