SPECIFICATIONS

Carotid

HS Transonic® Clinical Flowprobe
Specifications
Models:AU, FMC, FME, FMV, MB/MR, MBS/MRS, MU

USA/CanadaM

Transonic Systems Inc.
34 Dutch Mill Road
Ithaca, NY 14850
U.S.A.
Tel: +1 607-257-5300
Fax: +1 607-257-7256
support@transonic.com

PTransonic Europe B.V.
Business Park Stein 205
6181 MB Elsloo
The Netherlands
Tel: +31 43-407-7200
Fax: +31 43-407-7201
europe@transonic.com

Asia/PacificJapan

Transonic Asia Inc.
6F-3 No 5 Hangsiang Rd.
Dayuan, Taoyuan County
33747 Taiwan, R.O.C.
Tel: +886 3399-5806
Fax: +886 3399-5805
support@transonicasia.com

Nipro-Transonic Japan Inc.		
7th Floor, Maruha Building
11-1 Matsuba-cho
Tokorozawa City, Saitama
359-0044 Japan
Tel: +81 04-2946-8541
2797
Fax: +81 04-2946-8542
japan@transonic.com

AU-ISRT-FlowXLProbes-EN, Rev C Last Updated 10/26/23

CFlowXL® Clinical Flowprobe Specifications
Probe Size

Catalog # *# of Usesmm

Max FlowAccuracy

Max Zero Offset

Ultrasound Frequency

L/min% of Reading(ml/min)MHz

FMC-Series: Coronary steam sterilizable Probes (long handle, J-reflector)
1.5mm

HS1.5FMC-75750.5+/- 10+/- 24.82mm

HS2FMC-75750.5+/- 10+/- 24.83mm

HS3FMC-75751+/- 10+/- 32.44mm

HS4FMC-75752+/- 10+/- 6

2.4FMV-Series: Vascular steam sterilizable Probes (short handle, J-reflector)
1.5mm

HS1.5FMV-75

75

0.5+/- 10+/- 24.82mm

HS2FMV-75

75

0.5+/- 10+/- 24.83mm

HS3FMV-75

75

1+/- 10+/- 32.44mm

HS4FMV-75

752+/- 10+/- 62.46mm

HS6FMV-75

755+/- 10+/- 15

1.88mm

HS8FMV-757510+/- 10+/- 30

1.210mm

HS10FMV-757510+/- 10+/- 30

1.212mm

HS12FMV-75

7520+/- 10+/- 60

0.914mm

HS14FMV-75

7520+/- 10+/- 60

0.9

FME-Series: Carotid Endarterectomy steam sterilizable Probes (short handle, L-reflector)
1.5mm

HS1.5FME-75

75

0.5+/- 10+/- 24.82mm

HS2FME-7575

0.5+/- 10+/- 24.83mm

HS3FME-75

75

1+/- 10+/- 3

2.44mm

HS4FME-75

752+/- 10+/- 62.46mm

HS6FME-75

755+/- 10+/- 15

1.88mmHS8FME-75

7510+/- 10+/- 30

1.210mm

HS10FME-75

7510+/- 10+/- 30

1.2

MB-Series: Intracranial Charbel Micro-Flowprobes (long handle, single use)
1.5mm

HS1.5MB-01

10.2+/- 10+/- 3

4.82mm

HS2MB-01

10.5+/- 10+/- 5

3.63mm

HS3MB-01

11+/- 10+/- 10

2.4MBS-Series: Extracranial Micro-Flowprobes (short handle, single use)
3mm

HS3MBS-01

10.2+/- 10+/- 3

3.64mm

HS4MBS-01

10.5+/- 10+/- 6

2.46mm

HS6MBS-01

11+/- 10+/- 151.8Note: All HS Series Flowprobes have a cable length of 3 meters, from probe head to base of connector (+/- 5cm).

2AU-ISRT-FlowXLProbes-EN
Rev C

FlowXL® Clinical Flowprobe Specifications
Probe Size

Catalog # *# of Usesmm

Max Flow

Accuracy

Max Zero Offset

Ultrasound Frequency

L/min% of Reading(ml/min)MHz

MR-Series: Intracranial Charbel Micro-Flowprobes (long handle, multi-use)
1.5mm

HS1.5MR-16

160.2+/- 10+/- 34.82mm

HS2MR-16160.5+/- 10+/- 53.63mm

HS3MR-16161+/- 10+/- 102.4

MRS-Series: Extracranial Micro-Flowprobes (short handle, multi-use)
3mm

HS3MRS-16

160.2+/- 10+/- 33.64mm

HS4MRS-16

160.5+/- 10+/- 62.46mm

HS6MRS-16

161+/- 10+/- 15

1.8

MU-Series: Microvascular Probes (short handle, L-reflector)
0.7mm

HS0.7MU-50

50

0.05+/- 10+/- 0.4

9.61mmHS1MU-50

50

0.1+/- 10+/- 0.8

7.21.5mm

HS1.5MU-50

500.2+/- 10+/- 1.54.82mm

HS2MU-50

500.5+/- 10+/- 2.5

3.63mmHS3MU-50501+/- 10+/- 5

2.4AU-Series: COnfidence Flowprobes
Sizes 4&6 are 8-use, Sizes 8-36 are 75-use
4mm

HS4AU-08

82+/- 10+/- 8

7.26mm

HS6AU-08

85+/- 10+/- 20

4.88mm

HS8AU-75

7510+/- 10+/- 40

3.610mm

HS10AU-75

7510+/- 10+/- 40

3.612mm

HS12AU-75

75

20+/- 10+/- 80

2.414mm

HS14AU-75

75

20+/- 10+/- 80

2.416mm

HS16AU-75

75

50+/- 10+/- 200

1.820mm

HS20AU-75

75

50+/- 10+/- 200

1.824mm

HS24AU-7575100+/- 10+/- 400

1.2

28mm

HS28AU-75

75100+/- 10+/- 400

1.232mm

HS32AU-75

75200+/- 10+/- 800

0.9

36mm

HS36AU-75

75200+/- 10+/- 800

0.9* The suffix at the end of the Catalog Number (-01, -08, -16 or -75) represents the number of uses for that item.

AU-ISRT-FlowXLProbes-ENRev C3