Digital Blood Pressure Monitor
Model UA-1020-W
Instruction Manual Original

EnglishManuel d’instructions Traduction

Français

Manual de Instrucciones Traducción

EspañolManuale di Istruzioni Traduzione

Italiano

Instructiehandleiding Vertaling

Nederlands

Bruksanvisning Översättning

Svenska

Ohjekirja Käännös

Suomi

Instrukcja obsługi Tłumaczenie

Polski

使用手冊 翻譯1WMPD4002954D

中文

Talimat Kılavuzu ÇeviriTÜRKÇE‫ﺗﺭﺟﻣﻪ ﺩﻟﻳﻝ ﺍﻹﺳﺗﻌﻣﺎﻝ‬‫ﺗﺮﺟﻤﻪ ﺩﻓﺘﺮﭼﻪ ﺭﺍﻫﻨﻤﺎ‬‫ﻋﺭﺑﻲ‬‫ﻓﺎﺭﺳﯽ‬Contents
Dear Customers .................................................................... 2
Preliminary Remarks ............................................................. 2
Precautions ........................................................................... 2
Parts Identification ................................................................ 4
Symbols ................................................................................ 5
Operation Mode .................................................................... 7
Using the Monitor ................................................................. 8
Installing / Changing the Batteries ................................... 8Connecting the Air Hose .................................................. 9Connecting the AC Adapter .............................................. 9
Adjusting the Built-in Clock ........................................... 10
Selecting the Correct Cuff Size ....................................... 11
Applying the Arm Cuff ................................................... 11
How to Take Accurate Measurements ............................. 13
Measurement ................................................................. 13
After Measurement ........................................................ 13
Measurements .................................................................... 14
Normal Measurement ..................................................... 14
Measurement with the SET Pressure ............................... 15
Measurement with the Desired Systolic Pressure ............ 15
Notes for Accurate Measurement ................................... 16
TriCheckTM Measurement .................................................... 16
Recalling the Memory Data ................................................. 18
What Is The IHB/AFib Indicator? .......................................... 19
What Is The AFib? ............................................................... 19
Pressure Bar Indicator ......................................................... 19
WHO Classification Indicator ............................................... 19
About Blood Pressure.......................................................... 20
What is Blood Pressure? ................................................. 20
What is Hypertension and How is it Controlled? .............. 20
Why Measure Blood Pressure at Home?........................... 20
WHO Blood Pressure Classification ................................. 20
Blood Pressure Variations............................................... 20
Troubleshooting ................................................................. 21
Maintenance ....................................................................... 22
Technical Data .................................................................... 22

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure
monitor. Designed for ease of use and accuracy, this monitor will facilitate
your daily blood pressure regimen.
We recommend that you read through this manual carefully before
using the device for the first time.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity.
(0123: The reference number to the involved notified body)
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in the homehealthcare environment.
This device is designed to measure blood pressure and pulse rate of people
for diagnosis.

Precautions
Precision components are used in the construction of this device. Extremesin temperature, humidity, direct sunlight, shock or dust should be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff or storing the hose tightly twisted for long
periods, as such treatment may shorten the life of the components.
Take care to avoid accidental strangulation of babies or infants with thehose and cable.
Do not twist the air hose during measurement. This may cause injury due
to continuous cuff pressure.
The device and cuff are not water resistant. Prevent rain, sweat and waterfrom soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
Wireless communication devices, such as home networking devices, mobile
phones, cordless phones and their base stations, walkie-talkies can affect thisblood pressure monitor.
Therefore, a minimum distance of 30 cm should be kept from such devices.

English 2

Used equipment, parts and batteries are not treated as ordinary
household waste, and must be disposed of according to the applicable
local regulations.
When the AC adapter is used, make sure that the AC adapter can bereadily removed from the electrical outlet when necessary.
When reusing the device, confirm that the device is clean.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the arm must be squeezed by the cuff hardenough to temporarily stop blood flow through the artery. This may cause
pain, numbness or a temporary red mark to the arm. This condition willappear especially when measurement is repeated successively.
Any pain, numbness, or red marks will disappear with time.
Measuring blood pressure too frequently may cause harm due to blood flowinterference. Check that the operation of the device does not result in
prolonged impairment of blood circulation, when using the device repeatedly.
If you have had a mastectomy, please consult a doctor before using the device.
Do not let children use the device by themselves and do not use the device
in a place within the reach of infants. It may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
Unplug the AC adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially cause burns.
Allow the device to adapt to the surrounding environment before use(about one hour).
Clinical testing has not been conducted on newborn infants and pregnant
woman. Do not use on newborn infants or pregnant woman.
Do not touch the batteries, the DC jack, and the patient at the same time.
That may result in electrical shock.
Do not inflate without wrappping the cuff around the upper arm.

Contraindications
The following are precautions for proper use of the device.
Do not apply the cuff to an arm with another medical electrical equipment
attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult adoctor before using the device, to avoid medical problems.
Do not self-diagnose the measurement results and start treatment by yourself.
Always consult your doctor for evaluation of the results and treatment.
Do not apply the cuff on an arm with an unhealed wound.
Do not apply the cuff on an arm receiving an intravenous drip or bloodtransfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as ahigh-pressure oxygen chamber or an oxygen tent. It may cause a fire or explosion.
English 3

Parts Identification
DC Jack

Display

ButtonButtonSET Button

Air Connector Plug

START Button
Air Socket
Air Hose
Proper Fit Range

Artery Position MarkArm Cuff

Index Mark
Battery CoverBattery Compartment
1.5V Batteries(R6P, LR6 or AA)

Display

MEMORY

Average

Pressure Settings

IHB/AFib symbol
Cuff Fit Error Symbol

Systolic
Pressure

Movement Error Symbol
TriCheckTM Symbol

Diastolic
PressureHeart MarkWHO Classification Indicatorand Pressure Bar Indicator
Battery Indicator

Pulse Rate

AM / PM MarkEnglish 4

Clock Display

Symbols
Symbols that are printed on the device case and the AC adapter
Symbols

Function / Meaning

Recommended Action

Standby and Turn the device on.
Battery installation guide
Direct current
SN

Serial number
Date of manufacture
Type BF: Device, cuff and tubing aredesigned to provide special protection
against electrical shocks.
EC directive medical device label
WEEE label
Manufacturer
EU-representative
Polarity of DC jack

IP

International protenction symbol
Keep drySymbols that appear on the display
Symbols

Function / Meaning
Recommended Action
Appears while measurement is in progress. Measurement is in progress.
It blinks when the pulse is detected.
Remain as still as possible.
IHB/AFib symbol
Appears when an irregular heartbeat
is detected.
It may light when a very slight vibration
like shivering or shaking is detected.
TriCheckTM mode
Takes three consecutive measurements
automatically and displays the averagevalues of the three measurements.
Appears when a body or arm
movement is detected.

English 5

The reading may yield
an incorrect value. Take
another measurement.
Remain still during
measurement.

Symbols

Function / Meaning

Recommended Action

Appears during measurement when
the cuff is applied loosely.

The reading may yield anincorrect value. Apply the
cuff correctly, and takeanother measurement.

Previous measurements stored in MEMORY.
Average data
FULL BATTERY
The battery power indicator during
measurement.
LOW BATTERY
The battery power is low when it
blinks.
Unstable blood pressure due tomovement during measurement.
The systolic and diastolic values arewithin 10 mmHg of each other.
The pressure value did not increaseduring the inflation.
The cuff is not applied correctly.

Replace all batteries with newones when the mark blinks.
Try the measurement again.
Remain still during
measurement.

Fasten the cuff correctly,
and try the measurement
again.

PUL DISPLAY ERROR
The pulse is not detected correctly.
Blood pressure monitor internal error

SYS

Systolic blood pressure in mmHg

DIA

Diastolic blood pressure in mmHg

PUL

Pulse per minuteAM

Data taken between 4:00 and 9:59

PM

Data taken between 18:00 and 1:59
Pressure settings
Indicates the pressure value
previously set by the user.

English 6

Remove the batteries andpress the START button,
and then install the batteries
again. If the error still
appears, contact the dealer.

Operation Mode
1. Normal Measurement
Press the START button. Blood pressure is measured and data is stored
within the device. The device can store the last ninety sets of data inmemory automatically.

2. Recalling the Data
Press theor button to recall the data inmemory. The average of all measurements is
displayed, as indicated in the figure at the right.
Then, each time thebutton is pressed, thememory data is displayed as follows.
Average of all AM (morning) measurements
taken between 4:00 and 9:59.
Average of all PM (evening) measurements
taken between 18:00 and 1:59.
Most recent data (No.n, in the example,
No.35)
Last data (No.1)
For details on recalling the data, refer to thesection “Recalling the Memory Data”. (Referto page 18)

3. Deleting all Data Stored in Memory
Press both theand buttons. The
markand the battery indicator appear. Press and hold
both theand buttons until the illuminatedmark starts blinking to delete all data stored inmemory.

4. Measurement with the Desired Systolic Pressure
Refer to page 15 for measurement with the desired systolic pressure.
English 7

Using the Monitor
Installing / Changing the Batteries
1. Remove the battery cover.
2. Remove the used batteries andinsert new batteries into the battery
compartment as shown, taking
care that the polarities (+) and (-)
are correct.
Use only R6P, LR6 or AA batteries.
3. Attach the battery cover.

Step 1
Step 3

Step 1
Step 2

CAUTION
Insert the batteries as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(LOW BATTERY mark) blinks on the display, replace all batteries
with new ones. Do not mix old and new batteries. It may shorten the battery
life, or cause the device to malfunction.
Replace the batteries two seconds or more after the device turns off.
If
(LOW BATTERY mark) appears even after the batteries are replaced,make a blood pressure measurement. The device may then recognize the newbatteries.
(LOW BATTERY mark) does not appear when the batteries are drained.
The battery life varies with the ambient temperature and may be shorter at
low temperatures. Generally, four new LR6 batteries will last approximately
for one year when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device arefor testing monitor performance and may have a limited life.
Remove the batteries if the device is not to be used for a long time.
The batteries may leak and cause a malfunction.

English 8

Using the Monitor
Connecting the Air Hose

Air socket

Insert the air connector plug into
the air socket firmly.

Air connector plug

Connecting the AC Adapter
Insert the AC adapter plug into the
DC jack.
Next, connect the AC adapter to anelectrical outlet.

AC adapter plugDC jack
Use the specified AC adapter.
(Refer to page 23.)
When disconnecting the AC adapter from the electrical outlet, grasp andpull the AC adapter body out of the outlet.
When disconnecting the AC adapter plug from the blood pressure
monitor, grasp and pull the AC adapter plug out of the monitor.

English 9

Using the Monitor
Adjusting the Built-in Clock

Press

Adjust the clock prior to use.
1. Press the SET button until the year starts
blinking.
or button.
2. Select the year using thePress the SET button to set the current year
and move to month/day selection. The datecan be set anywhere between the years 2010
and 2059.
3. Select the month using theor button.
Press the SET button to set the current
month and move to day selection.
or button.
4. Select the day using thePress the SET button to set the current day
and move to hour/minute selection.
5. Select the hour using theor button.
Press the SET button to set the current hourand move to minute selection.
or button.
6. Select the minute using the(Press the SET button to go to the pressure
setting mode. For details, refer to page 15.)
Press the START button to turn the device off.

Year

or
Month

Day

or

or
Hour

MinuteNote: After three minutes of non-operation, the device
will turn off automatically.
When the clock has not been
Turn off
set, the clock display indicatesdashes as shown to the right.
Pressing the START
button will turn theHolding down theorbuttonwill change the value continuously.
device off anytime.

English 10

oror

Pressuresetting mode

Using the Monitor
Selecting the Correct Cuff Size
Using the correct cuff size is important for an accurate reading. If the cuff is
not the proper size, the reading may yield an incorrect blood pressure value.
The arm size is printed on each cuff.
The index
and proper fit range, on the cuff, tell you if you are applying thecorrect cuff. (Refer to Table "Symbols that are printed on the cuff" on the next
page)
If the index
points outside of the range, contact your local dealer topurchase a replacement cuff.
The arm cuff is a consumable. If it becomes worn, purchase a new one.
Arm Size
Recommended Cuff Size
31 cm to 45 cm
Large Adult Cuff
22 cm to 42 cm
Wide Range Cuff
22 cm to 32 cm
Adult Cuff
16 cm to 24 cm
Small Adult Cuff
Arm size: The circumference at the biceps.

Catalog Number
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Applying the Arm Cuff
1. Wrap the cuff around the upper arm,
about 1 to 2 cm above the inside ofthe elbow, as shown.
Place the cuff directly against theskin, as clothing may cause a faintpulse, and result in a measurement
error.

Do not roll up shirtsleeve tightly.
Cuff

Air Hose
Artery position mark2. Constriction of the upper arm,
caused by tightly rolling up a
shirtsleeve, may prevent accurate
readings.
3. Confirm that the index
within the proper fit range.

1-2 cmSecure the fabricfastener firmly.

pointsIndex

English 11

and proper fit range

Using the Monitor
Symbols that are printed on the cuff
Symbols
Function / Meaning

Recommended Action
Set themark on theartery of the upper arm orin line with the ring finger
on the inside of the arm.

Artery position mark
▲
REF

Index
Catalog number
Proper fit range for the adult cuff.
It's printed on the adult cuff.
Use the large adult cuff
Over range printed on the adult cuff
instead of the adult cuff orand wide range cuff.
wide range cuff.
Proper fit range for the wide range cuff.
It's printed on the wide range cuff.
Proper fit range for the large adult cuff.
It's printed on the large adult cuff.
Under range printed on the adult cuff Use the small adult cuff
instead of the adult cuff orand wide range cuff.
wide range cuff.
Proper fit range for the small adult cuff.
It's printed on the small adult cuff.
Use the adult cuff instead
It's printed on the large adult cuff andof the large adult cuff orsmall adult cuff.
small adult cuff.
Lot number
Proper fit range

Large adult cuff

A

L

Wide range cuff

S

W

L

Adult cuff

S

A

L

S

A

Small adult cuff

English 12

Using the Monitor
How to Take Accurate Measurements
For the most accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross yourlegs. Keep your feet flat on the floor and straighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same level as your heart.
Remain still and keep quiet during measurement.
Do not measure immediately after physical exercise or a bath. Rest fortwenty or thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

Measurement
During measurement, it is normal for the cuff to feel very tight. (Do not be
alarmed).

After Measurement
After measurement, press the START button to turn the device off.
Remove the cuff and record your data. After one minute of non-operation,
the device will turn off automatically.

English 13

Measurements
Before measurement, read the section “Notes for Accurate Measurement”
on page 16.

Normal Measurement
1.

Place the cuff on the arm (preferably the left arm).
Sit quietly during measurement.

2.

Press the START button.
Press
All of the display segments are displayed.
Zero (0) is displayed blinking briefly.
The display changes, as indicated in the
All of the display
segments
displayedfigure at the right, as the measurement
begins. The cuff starts to inflate. It is normal
for the cuff to feel very tight. A pressure bar
indicator is displayed, on the left edge of thedisplay, during the inflation.
Zero display
Note: If you wish to stop inflation at any time,
Starts inflationpress the START button again.

3.

When inflation is complete, deflation starts
automatically and(heart mark) blinks,
indicating that the measurement is inprogress. Once the pulse is detected, themark blinks with each pulse beat.
Note: If an appropriate pressure is not obtained, thedevice starts to inflate again automatically.
To avoid re-inflation, see thesection “Measurement with the
SET Pressure” on the next page.

4.

When the measurement is complete, thesystolic and diastolic pressure readings
and pulse rate are displayed.
The cuff exhausts the remaining air anddeflates completely.

5.

Press the START button to turn the device off.
After one minute of non-operation,
the device will turn off automatically.

Note: Allow at least three minutes between
measurements on the same person.

English 14

At heart level
START

Pressurizing

Measurementin progress
Systolic pressure
Diastolic pressure
WHO classification
Pulse rate
Exhausts remaining
air automaticallyMeasurements
The UA-1020-W is designed to detect the pulse and to inflate the cuff to asystolic pressure level automatically.
If re-inflation occurs repeatedly, use the following methods.

Measurement with the SET Pressure
During the blood pressure measurement, re-inflation may occur.
A fixed pressure value can be set to avoid re-inflation.
1.

2.

At step 6 of the clock adjustment procedure(see page 10), press the SET button to go to
the pressure setting mode. The current
setting blinks.

orPress theorbutton to select a pressure
value about 30 mmHg or more above yourexpected systolic pressure from the following.
AUTO : Automatic pressurization (default value)
180 : Pressure value of 180 mmHg (fixed)
210 : Pressure value of 210 mmHg (fixed)
Turn off
240 : Pressure value of 240 mmHg (fixed)

3.

TriChecksetting mode

Press the SET button to go to the TriCheck setting mode.
Press the START button to turn the device off. After three minutes of
non-operation, the device will turn off automatically.
The next measurement will be performed with the new pressure value.

Measurement with the Desired Systolic Pressure
Use this method when re-inflation occurs repeatedly even if the pressure value
is set to 240 in the procedure above or when the results are not displayedeven if the pressure decreases to 20 mmHg or less.
1.

Place the cuff on the arm (preferably the left arm). At heart level2.

Press and hold the START button until a
number about 30 to 40 mmHg higher
than your expected systolic pressure
appears.

3.

When the desired number is reached,release the START button to startmeasurement. Continue to measure
your blood pressure as described on
the previous page.
English 15

Press and hold the button

START

Release the button
at the desired
systolic pressure
Refer to the previous pagefor measurement

Notes for Accurate Measurement

Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five to ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.

Should the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. Refer to page 6 for thedescription of symbols.
This blood pressure monitor is intended for use by adults. Consult withyour physician before using this device on a child. A child should not use
this device unattended.
The automatic blood pressure monitor’s performance may be affected byexcessive temperature or humidity, or altitude.

TriCheckTM Measurement
The TriCheckTM mode takes three consecutive measurements automaticallyand displays the average values of the three measurements.

Selecting the TriCheckTM ModeIn the pressure setting mode,1.

In the pressure setting mode, press
the SET button to enter the
TriCheckTM setting mode. The current
setting blinks.

2.

Press theorbutton to turn theTM
TriCheck mode ON or OFF.
The current setting blinks
ON: TriCheckTM mode
OFF: Normal measurement mode (default value)or

3.

Press the START or SET button toturn the device off. After threeminutes of non-operation, the device
will turn off automatically.

English 16

press

Change the setting
or START

Measurement Using the TriCheckTM Mode1.

2.

Press the START button.
All of the display segments are
displayed.
Zero (0) is displayed blinking briefly
and the first measurement starts.
When the measurement is complete,
the systolic and diastolic pressure
readings and pulse rate are displayed,then a one-minute countdown starts forthe second measurement.

3.

After one minute, the second
measurement starts.

4.

When the measurement is complete,
the average readings of the first andsecond measurements are displayed,then a one-minute countdown starts forthe third measurement.

5.
6.

At heart level
START

Press
All of the display
segments displayed
First measurement

One minute after

After one minute, the third
measurement starts.

Results of thefirst measurement

Second measurement

When the measurement is complete,
the average readings of the threemeasurements are displayed andstored in memory.

Notes:
During the measurement, theTriCheckTM symbol is displayed.
To cancel the measurement, press
the START button. In this case, nodata is stored in memory.
When the cuff fit error symbol appears
during the first measurement, cancel
the measurement, reapply the cuff
correctly and start a new measurement.
After measurement, the averagedata of the three measurements isstored in memory.
No data will be stored, whencancelingthemeasurements,
before completion of the third
measurement.
English 17

One minute after

Average of the first andsecond measurements

Third measurement

Average of thethree measurements

Recalling the Memory Data
Note: This device stores the last ninety sets of data in memory automatically.
or button.
1. Press the
The average of all measurements andthe number of data are displayed.
(If no data, “0” is displayed. Press the, or START button to turn thedevice off.)Press

2. Each time thebutton (or the
button to display the data in thereverse order) is pressed, the memory
data is displayed as follows.
Average of all AM (morning) measurements
taken between 4:00 and 9:59.
(In the example, 10 measurements. If
no data, “--” is displayed.)or
Average systolic
Average diastolic
Average pulse

Average of
all the data

Averageof AM data

Average of all PM (evening) measurements
taken between 18:00 and 1:59.
(In the example, 9 measurements. If
no data, “--” is displayed.)
Most recent data (No.n, in theexample, No.35)
Three seconds after the data number
display, the measurement data is displayed.

Last data (No.1)
Three seconds after the data number
display, the measurement data isdisplayed.

Average systolic
Average diastolic
Average pulse

Average systolic
Average diastolic
Average pulse
Averageof PM data
Systolic
Diastolic
Pulse
Most recent data

3. After the last data is displayed, press
thebutton to return the average
display of all measurements.
4. Press the START button to turn thedevice off. After one minute of
non-operation, the device will turn off
automatically.
English 18

Systolic
Diastolic
Pulse
Last data (Oldest)

What Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements,
the IHB/AFib indicator will appear on the display with the measurement
values.
Note: We recommend contacting your physician if you see thisIHB/AFib indicator frequently.

What Is The AFib?
The heart contracts due to electrical signals occurring in heart and sends
blood through the body. Atrial fibrillation (AFib) occurs when the electricalsignal in the atrium becomes confused and leads to disturbances in thepulse interval. AFib can cause blood to stagnate in the heart, which caneasily create clots of blood, a cause of stroke and heart attack.

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing
Releasing air

Inflation in progress

Measurement in progress
Inflation complete

WHO Classification Indicator
Each segment of the bar indicator
corresponds to the WHO blood pressureclassification described on the next page.
WHO Classification Indicator
Severe hypertension
Moderate hypertension
Mild hypertension
High normal
NormalOptimal: The indicator displays a segment, basedon the current data, corresponding to the
WHO classification.
English 19

Example:Moderate hypertension

Mild hypertension

High normal

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of the arteries.
Systolic pressure occurs when the heart contracts. Diastolic pressure occurs
when the heart expands. Blood pressure is measured in millimeters of
mercury (mmHg). One's natural blood pressure is represented by thefundamental pressure, which is measured first thing in the morning while one
is still at rest and before eating.

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left unattended,
can cause many health problems including stroke and heart attack.
Hypertension can be controlled by altering lifestyle, avoiding stress, and withmedication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?
Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

WHO Blood Pressure Classification

Blood Pressure Variations

Diastolic blood pressureStandards to assess high Reference Material: Journal of Hypertension 1999, Vol 17 No.2
Grade 3 hypertension (severe)
blood pressure, without mmHg
110
regard to age, have been
Grade 2 hypertension (moderate)105
established by the World
100
Health Organization (WHO),
Grade 1 hypertension (mild)95
as shown in the chart.
90
85
80

High-normal

NormalOptimalAn individual’s blood pressure
120 130 140 150 160 170 180
varies greatly on a daily andSystolic blood pressure
mmHg
seasonal basis. It may varyby 30 to 50 mmHg due to various conditions during the day. In hypertensive
individuals, variations are even more pronounced. Normally, the bloodpressure rises while at work or play and falls to its lowest levels during sleep.
So, do not be overly concerned by the results of one measurement.
Take measurements at the same time every day using the procedure

English 20

Typical fluctuation within a day(Measured every five minutes)
Systolic
Blood pressure

described in this manual to get to know yournormal blood pressure. Regular readings givea more comprehensive blood pressure history.
Be sure to note the date and time whenrecording your blood pressure. Consult yourdoctor to interpret your blood pressure data.

Sleep

Diastolic
PM

AM

Time

Troubleshooting
ProblemPossible Reason
Batteries are drained.
Nothingappears on thedisplay, even
Battery terminals are notwhen the device in the correct position.
is turned on.
Battery voltage is too low.
(LOW BATTERY mark)The cuff does
blinks. If the batteries are
not inflate.
drained completely, themark does not appear.
The cuff is not applied
properly.
You moved your arm or
body during
The device does measurement.
not measure.
Readings are
The cuff position is not
too high or too correct.
low.

Other

The value is different
from that measured at aclinic or doctor’s office.

Recommended Action
Replace all batteries with new ones.
Reinstall the batteries withnegative and positive terminals
matching those indicated on thebattery compartment.
Replace all batteries with new
ones.
Apply the cuff correctly.
Make sure you remain still andquiet during measurement.
Sit comfortably and still.
Place your arm on a table with yourpalm facing upward and the cuff at
the same level as your heart.
If you have a very weak or
irregular heartbeat, the device
may have difficulty in determining
your blood pressure.
Refer to “Why Measure Blood
Pressure at Home?”.

Remove the batteries. Placethem back properly and takeanother measurement.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this product, as any attempt todo so will make your warranty invalid.
English 21

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem using
the troubleshooting instructions, contact the authorized dealer in your areaor our customer service department. The A&D customer service will
provide technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life.
However it is generally recommended to have the device inspected every
2 years, to ensure proper functioning and accuracy. Please contact theauthorized dealer in your area or A&D for maintenance.

Technical Data
Type

UA-1020-W

Measurement method

Oscillometric measurement

Measurement range

Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / minute

Measurement accuracy

Pressure: ±3 mmHg
Pulse:±5%

Power supply

4 x 1.5V batteries (R6P, LR6 or AA) or
AC adapter (TB-233C) (Not included)

Number of measurements

Approx. 1000 times LR6 (alkaline batteries)
Approx. 300 times R6P (manganese batteries)
With pressure value of 180 mmHg at room
temperature of 23°C

Classification

Internally powered ME equipment (Supplied bybatteries) /Class II (Supplied by adapter)
Continuous operation mode

Clinical test

According to ISO81060-2：2013
In the clinical validation study, K5 was used on 85
subjects for determination of diastolic blood

EMD

IEC 60601-1-2: 2014

MemoryLast 90 measurements

Operating conditions+10 to +40 °C / 15 to 85 %RH / 800 to 1060 hPa
English 22

Transport / Storage conditions -20 to +60 °C / 10 to 95 %RH / 700 to 1060 hPa
Dimensions
Approx. 140 [W] x 60 [H] x 105 [D] mm
WeightApprox. 285 g, excluding the batteries

Applied part

Cuff

Ingress protection

Device: IP20

Useful life

Device: 5 years (when used six times a day)Cuff: 2 years (when used six times a day)AC adapter: 5 years (when used six times a day)Accessory AC adapter

The adapter is to connect the blood pressure
monitor to a power source at home.

TB-233C

Type BF

Please contact your local A&D dealer forpurchasing.
The AC adapter is required to be inspected or
replaced periodically.
Symbols that are printed on the AC adapter
Symbols
Function / Meaning
For indoor use only
Class II device
Thermal fuse
Fuse
EC directive device label
EAC certification device label
Polarity of AC adapter plug

Accessories sold separately
Cuff
Catalog Number
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Cuff Size
Large adult cuff
Wide range cuff
Adult cuff
Small adult cuff

Catalog Number
TB-233C

Plug
Type CArm Size
31 cm to 45 cm
22 cm to 42 cm
22 cm to 32 cm
16 cm to 24 cm

AC adapter

Note: Specifications are subject to change without prior notice.
IP classification is the degrees of protection provided by enclosures inaccordance with IEC 60529. This device is protected against solid foreign
objects of 12 mm diameter and greater such as a finger. This device is notprotected against water.
English 23

MEMO

English 24

Sommaires
Chers clients ......................................................................... 2
Remarques préliminaires ...................................................... 2
Précautions ........................................................................... 2
Nomenclature ....................................................................... 4
Symboles .............................................................................. 5Mode opératoire ................................................................... 7
Utilisation de l’appareil ......................................................... 8
Mise en place / Remplacement des piles...................................... 8
Branchement du tuyau d’air .......................................................... 9
Branchement de l’adaptateur secteur .................................................. 9
Réglage de l’horloge de l’appareil ............................................... 10
Choix du brassard ....................................................................... 11
Mise en place du brassard .......................................................... 11
Comment faire des mesures correctes .......................................... 13
Pendant la mesure ...................................................................... 13
Après la mesure .......................................................................... 13
Les mesures ....................................................................... 14
Mesure normale .......................................................................... 14
Mesure avec la pression configurée ........................................... 15
Mesure avec la pression systolique désirée ............................... 15
Consignes et observations pour une mesure correcte ................ 16
Mesure TriCheckTM .............................................................. 16
Rappel des données mémorisées ........................................ 18
Qu’est ce que l’indicateur de RCI/FA? ................................. 19
Qu’est-ce que la FA? ........................................................... 19
Indicateur de progression du gonflage ................................ 19
Indicateur de tension selon la classification OMS............................ 19
Pression sanguine ............................................................... 20
Qu'est-ce que la pression sanguine? .......................................... 20
Qu'est-ce que l'hypertension et comment la contrôler? .............. 20
Comment mesurer la pression sanguine chez soi? .................... 20
Classification de la tension selon l’OMS ..................................... 20
Variations de la pression sanguine ............................................. 20
Résolution des problèmes ................................................... 21
Entretien ............................................................................. 22
Fiche technique .................................................................. 22

Français

1

Chers clients
Félicitations pour l’achat de ce tensiomètre électronique A&D de dernière
génération. Conçu pour faciliter son utilisation et augmenter sa précisionil vous donnera au quotidien des mesures précises de la pression
sanguine.
Lisez ce manuel entièrement et soigneusement avant d’utiliser
l’appareil pour la première fois.

Remarques préliminaires
Cet appareil est conforme à la directive européenne 93/42 CEE relative
aux dispositifs médicaux et porte le marquage. (0123 fait référence
à l’organisme notifié ayant certifié le produit)
Cet appareil est conçu pour être utilisé sur les adultes, et non sur lesnouveau-nés ou les enfants.
Lieu d’utilisation. Cet appareil est destiné à être utilisé par vous-même
dans un environnement domestique de soins de santé.
Cet appareil est conçu pour mesurer la tension artérielle et la fréquencecardiaque des personnes dans un objectif de diagnostic.

Précautions
Cet appareil est équipé de pièces de précision. C'est pourquoi il fautéviter de l’exposer à de fortes températures, une humidité extrême et ausoleil. De même il faut éviter les chocs et le protéger de la poussière.
Nettoyez l’appareil et le brassard avec un chiffon doux et sec, ou un
chiffon légèrement humidifié avec de l’eau et un détergent neutre.
N’utilisez jamais d’alcool, du benzène, de diluant ou autres produits
chimiques pour nettoyer l’appareil ou le brassard.
Evitez de tordre le brassard ou de laisser le tuyau trop plié pendant de longues
périodes ce qui pourrait avoir une influence néfaste sur la durée de vie des pièces.
Veillez à ce que les bébés et les enfants en bas âge ne s’étranglent pas avec letuyau et le câble.
Ne tordez pas le tuyau d’air pendant la mesure. La pression continue du brassard
pourrait vous blesser.
L'appareil et le brassard ne sont pas imperméables. Evitez de les mouiller
avec de l'eau, de la transpiration ou la pluie.
Il est possible que les mesures soient altérées si l’appareil est utilisé à
proximité de téléviseurs, de fours à micro-ondes, de téléphones cellulaires,de sources de rayonnement X ou d’autres appareils émettant de forts
champs électriques.
Les dispositifs de communication sans fil, tels que les dispositifs réseau à
domicile, les téléphones mobiles, les téléphones sans fil et leurs stations de base,les talkies-walkies peuvent affecter ce moniteur de pression sanguine. Par
conséquent, vous devez maintenir une distance minimale de 30 cm par rapport àces dispositifs.
L'appareil usé, les pièces et les piles ne doivent pas être jetées comme lesdéchets ménagers ordinaires, elles doivent être traitées conformément à lalégislation en vigueur.
Lorsque vous utilisez l'adaptateur secteur, assurez-vous qu'il peut êtrefacilement retiré de la prise électrique si nécessaire.
Lorsque vous réutilisez l'appareil, vérifiez qu'il est propre.
Français2

Ne modifiez pas l'appareil. Cela pourrait l'endommager ou provoquer desaccidents.
Pour mesurer la tension artérielle, le bras doit être suffisamment pressé par lebrassard afin de bloquer temporairement la circulation sanguine dans l'artère.
Cela peut provoquer une douleur, un engourdissement ou une marque rouge
temporaire sur le bras. Cet état survient particulièrement si la mesure est
répétée de manière successive.
Les douleurs, les engourdissements ou les marques rouges disparaissent
avec le temps.
Une mesure trop fréquente de la pression sanguine peut nuire au flux sanguin.
Si vous utilisez l’appareil à plusieurs reprises, assurez-vous que l’utilisation del’appareil n’entrave pas la circulation sanguine de manière prolongée.
Veuillez consulter un médecin avant d’utiliser cet appareil si vous avez subi
une mastectomie.
Ne laissez pas d’enfants utiliser l’appareil par eux-mêmes et n’utilisez pas
l’appareil dans un endroit où il se trouve à portée d’enfants en bas âge. Cela
pourrait provoquer des accidents ou des dommages.
L’appareil comporte de petites pièces qui risqueraient d’étouffer les enfants enbas âge s’ils venaient à les avaler.
Débranchez l’adaptateur secteur lorsqu’il n’est pas utilisé pendant la mesure.
L’utilisation d’accessoires non décrits dans ce manuel peut compromettre lasécurité.
Si la pile est en court-circuit, elle peut devenir chaude et causer potentiellement
des brûlures.
Laissez l’appareil s’adapter à l’environnement ambiant avant utilisation(environ une heure).
Aucun essai clinique n’a été mené sur les nouveau-nés et les femmes enceintes.
N’utilisez pas l’appareil sur des nouveau-nés ou des femmes enceintes.
Ne touchez pas les piles, la prise CC et le patient en même temps. Cela
pourrait entraîner un choc électrique.
Ne gonflez pas sans avoir enroulé le brassard autour de la partie supérieure du bras.

Contre-indications

Vous trouverez ci-après des précautions destinées à l'utilisation appropriée del'appareil.
N'appliquez pas le brassard sur un bras sur lequel un autre équipement
médical électrique est fixé. L'équipement pourrait ne pas fonctionnercorrectement.
Les personnes victimes d'une insuffisance circulatoire grave au niveau dubras doivent consulter un médecin avant d'utiliser l'appareil afin d'éviter
tout problème médical.
N'effectuez pas vous-même de diagnostic des résultats de la mesure etne commencez pas de traitement vous-même. Consultez toujours votre
médecin pour une évaluation des résultats et du traitement.
N'appliquez pas le brassard sur un bras portant une blessure non guérie.
N'appliquez pas le brassard sur un bras recevant un goutte-à-goutte
intraveineux ou une transfusion sanguine. Cela pourrait provoquer desblessures ou des accidents.
N'utilisez pas l'appareil en présence de gaz inflammables tels que desgaz anesthésiants. Cela pourrait provoquer une explosion.
N'utilisez pas l'appareil dans des environnements à forte concentrationd'oxygène, tels qu'une chambre à oxygène sous haute pression ou une
tente à oxygène. Cela pourrait provoquer un incendie ou une explosion.

Français3Nomenclature
Connecteur decourant continu
BoutonBoutonBouton REGLAGES

Affichage

Prise de connexion d’air

Bouton START(mise en marche)

Connecteur d’air
Tuyau d’air
Plage d’accord
appropriéeMarque deposition de l’artère
Brassard

Marque d’index
Couvercle logement des piles
Logement des piles

Piles 1,5 V(R6P, LR6 ou AA)

Partie d’Affichage MEMOIRE

MoyenneRéglages de la pression

Symbole de RCI/FA

Symbole d’erreur de fixation du brassard

PressionsystoliqueSymbole d’erreur de mouvement
Symbole TriCheckTM

Pression
diastoliqueSymbole du cœurIndicateur de la Classification
suivant l’OMS et Indicateur deprogression du gonflage

Pouls

Indicateur des piles
Marque AM / PM
Français

Affichage de l’horloge
4Symboles
Symboles imprimés sur l’appareil et sur l’adaptateur secteur
SymbolesSN

Fonction / Signification

Action recommandée

Pour allumer ou éteindre l’appareil
Repères pour installer les piles
Courant continu
Numéro de série
Date de fabrication
Type BF: L’appareil, le brassard et les
tubes isolants ont été conçus de manière
à fournir une protection particulière
contre les chocs électriques.

Dispositif Médical conforme à ladirective européenne 93/42 CEE
Déchets d’Equipements Electriques etElectroniques (DEEE)IP

Ne pas jeter avec les ordures
ménagères. Utilisez les filières
mises à votre disposition.

FabricantMandataire Européen
Consultez le brochure d'instructions
Polarité du connecteur de courant continu
Symbole de protection internationale
Conserver au sec

Symboles affichés à l'affichage
SymbolesFonction / Signification

Action recommandée

Cet indicateur apparaît lorsque la mesure L’appareil est en train deest en cours. Il clignote quand lesmesurer; restez calme.
pulsations cardiaques sont détectées.
Symbole de RCI/FA
Cet indicateur apparaît en cas de détectiond’un rythme cardiaque irrégulier.
Il est susceptible de s’allumer lors de ladétection d’une très faible vibration telle
qu’un frissonnement ou un tremblement.
Le mode TriCheckTM prend
automatiquement trois mesuresconsécutives et affiche les valeursmoyennes de celles-ci.
La lecture peut être àl’origine d’une valeur
Symbole affiché lors de la détection
erronée. Recommencez lad'un mouvement du corps ou du bras. mesure. Restez parfaitement
tranquille pendant la mesure.
Français

5Symboles

Fonction / Signification

Action recommandéeSymbole affiché pendant la mesure si
le brassard n’est pas assez serré.

La lecture peut être àl’origine d’une valeur
erronée. Attachez lebrassard correctement, etrecommencez la mesure.

Les mesures précédentes conservées
dans la MEMOIRE.
Données moyennes

BATTERIE CHARGÉE

Indicateur de tension des piles pendant
la mesure.

BATTERIE FAIBLE

SYS
DIA
PUL
AM
PM

Remplacer toutes les piles
La tension des piles est faible quand il par des neuves quand cet
indicateur clignote.
clignote.
Apparaît en cas de fluctuation de la
Recommencez la mesure.
pression due à un mouvement pendant Restez parfaitement tranquillela mesure.
pendant la mesure.
Apparaît si la difference entre lapression systolique(tension maxima) etla pression diastolique(tension minima) est
inférieure à 10 mmHg.
Apparaît lorsque la pression n’augmente
Attachez le brassard
pas pendant le gonflage du brassard.
correctement, etApparaît lorsque le brassard n’est pas recommencez la mesure.
attaché correctement.
ERREUR D’AFFICHAGE DU POULS
La pression n’est pas détectéecorrectement.
Retirez les piles et appuyez
sur le bouton START, puis
Erreur interne du moniteur de laremettez les piles en place.
pression sanguine
Si l’erreur s’affiche encore,contactez votre revendeur.
Pression systolique en mmHg.
Pression diastolique en mmHg.
Pouls: nombre de pulsations cardiaques par
minute.
Données mesurées entre 4h00 et 9h59
Données mesurées entre 18h00 et 1h59
Réglages de la pression
Indique la valeur de la pression
précédemment selectionnée par l’utilisateur.
Français6Mode opératoire
1. Mesure normale avec les données conservées
Appuyez sur le bouton START. La pression sanguine est mesurée et lesdonnées sont conservées dans l’appareil. Cet appareil peut conserver les
90 dernières données automatiquement dans la MEMOIRE.

2. Rappel des données en mémoireoupour rappeler les
Appuyez sur le bouton
données.
La moyenne des données de toutes les mesures est
affichée, comme indiqué sur la figure de droite.
, affiche

Ensuite, chaque pression sur le bouton
les données mémorisées comme suit.

Moyenne de toutes les mesures AM (du matin)
prises entre 4h00 et 9h59.
Moyenne de toutes les mesures PM (du soir)prises entre 18h00 et 1h59.
Données les plus récentes (N°35 dans l’exemple)
Dernières données (N° 1)
Pour plus de détails sur le rappel des données,
reportez-vous à la section « Rappel des donnéesmémorisées ». (Voir la page 18.)

3. Suppression de toutes les données en mémoire
Appuyez sur les deux boutons
et . La marqueetl'indicateur des piles s'affichent. Appuyez sur les deux
boutons
et , et maintenez-les enfoncés jusqu'à ce quela marque éclairée
commence à clignoter poursupprimer toutes les données stockées en mémoire.

4. Mesure avec la pression systolique désirée
Voir la page 15 pour la mesure avec la pression systolique désirée.
Français

7

Utilisation de l’appareil
Mise en place / Remplacement des piles
1. Enlevez le couvercle du logement despiles.
2. Enlevez des piles usées et insérez
des piles neuves dans le logement en
respectant bien les polarités (+ et -)indiquées.
Utilisez uniquement des piles R6P,
LR6 ou AA.
3. Fermez le couvercle du logement des piles.

Etape 1

Etape 3

Etape 1

Etape 2ATTENTION
Veillez à insérer les piles dans le bon sens, sinon l’appareil ne fonctionnera
pas.
Quand le symbole(piles faibles) clignote sur l’écran, l’appareil indiqueque les piles doivent être remplacées. Changer alors toutes les piles pardes neuves. Ne mélangez pas les piles neuves et anciennes car cela
pourrait raccourcir leur durée de vie ou détériorer l’appareil.
Remplacez les piles au moins deux secondes après avoir éteint l'appareil.
Si le symbole(piles faibles) s'affiche même après le remplacement
des piles, réalisez une mesure de la pression sanguine. L'appareil devrait
alors reconnaître les nouvelles piles.
Le symbole(piles faibles) n'apparaît pas lorsque les piles sontépuisées.
La durée de vie des piles varie avec la température ambiante et peut êtreraccourcie par des températures basses. En général, quatre piles neuves
LR6 durent approximativement un an en cas de mesure deux fois par jour.
Utilisez uniquement les piles spécifiées. Les piles fournies avec l'appareil sontpour tester les performances de l'écran d’affichage et peuvent donc avoir une
courte durée de vie.
Retirez les piles lorsque l'appareil n'est pas utilisé pendant un long moment.
Les piles pourraient couler ce qui entraînerait un mauvais fonctionnement
de l'appareil.

Français

8

Utilisation de l’appareil
Branchement du tuyau d’air

Connecteur d’air

Insérez fermement la prise deconnexion d’air dans le connecteur d’air.

Prise de connexion d’air

Branchement de l’adaptateur secteur

Fiche de l'adaptateur
secteurIntroduire la fiche de l'adaptateur secteurdans le connecteur de courant continu.
Brancher ensuite l’adaptateur sur une prisede courant.

Connecteur de courant
continu

Utilisez l'adaptateur secteur spécifié.
(Voir la page 23.)
Lorsque vous débranchez l’adaptateur secteur de la prise secteur, tenez
le corps de l’adaptateur et débranchez-le de la prise.
Lorsque vous débranchez l’adaptateur secteur du moniteur de pression
sanguine, tenez la fiche de l’adaptateur et débranchez-la du moniteur.

Français

9

Utilisation de l’appareil
Réglage de l’horloge de l’appareil

AppuyezRégler l’horloge avant d’utiliser l’appareil.

ou
Année

1. Appuyez sur le bouton REGLAGES jusqu’à ce
que les nombres commencent à clignoter.
2. Choisissez l’année en utilisant le bouton
ou .
Appuyez sur le bouton REGLAGES pour valider lechoix et passer au réglage du mois. Vous pouvez
définir n’importe quelle date entre 2010 et 2059.

ou
Mois

ou .
3. Choisissez le mois en utilisant le bouton
Appuyez sur le bouton REGLAGES pour valider lechoix et passer au réglage du jour.

ou
Jour

4. Choisissez le jour en utilisant le bouton
ou .
Appuyez sur le bouton REGLAGES pour valider lechoix et passer au réglage de l’heure.
ou .
5. Choisissez l’heure en utilisant le bouton
Appuyez sur le bouton REGLAGES pour valider
le choix et passer au réglage des minutes.

ou
Heure

ou .
6. Choisissez la minute en utilisant le bouton
Appuyez sur le bouton REGLAGES pour valider lechoix et passer au réglage de la pression. Pour plusde détails, reportez vous à la page 15.)
Appuyez sur le bouton START pour éteindre l’appareil.
Note: L'appareil s’éteint automatiquement
après trois minutes sans utilisation.
Si l'horloge n'a pas été réglée,des tirets sont affichés comme
indiqué à droite.
Maintenir le bouton
ouenfoncépermet de changer la valeur en continu.

Français

Minuteou

Éteindre
Mode de réglage
Appuyer sur le bouton
START permet d'éteindre de la pression.
l'appareil à tout moment.

10

Utilisation de l’appareil
Choix du brassard
L’utilisation d’un brassard adapté à la morphologie du patient est importante pourun résultat précis. Si la taille du brassard n’est pas appropriée il peut en résulterl’affichage de valeurs erronées.
La taille du brassard (circonférence du biceps) est imprimée sur chaque brassard.
L’indexet la plage d’accord appropriée sur le brassard vous indiquent si vous
mettez le brassard correctement. (Reportez-vous à « Symboles imprimés sur lebrassard », à la page suivante.)
Si l’index
se trouve en dehors de la plage, prière de prendre contact avec
votre revendeur local afin d’acheter le brassard pour le remplacement.
Le brassard est un produit consommable. S’il est usé, achetez-en un neuf.
Taille du bras
Taille du brassard recommandée
31 cm à 45 cm
Brassard adulte de grande taille
22 cm à 42 cm
Brassard de grande taille
22 cm à 32 cm
Brassard adulte
16 cm à 24 cm
Brassard adulte de petite taille
Taille du bras : circonférence au niveau du biceps.

Mise en place du brassard

Ne remontez pas la mancheétroitement.

1. Enroulez le brassard autour du bras,à 1 ou 2 cm au-dessus de l’intérieur
du coude, comme indiqué surl’illustration ci-contre. Placez lebrassard directement sur la peau,
car un vêtement pourrait assourdir
la pulsation et provoquer une erreurde mesure.

Brassard
1-2 cmTuyau d’air
Marque de position de l’artère

2. Evitez également de remonterétroitement la manche, car ellecomprimerait le bras ce qui pourrait
aussi provoquer une erreur demesure.

Veillez à bien fixer
l'attache à tissu.

se trouve
3. Confirmez que l’index
à l’intérieur de la plage d’accord
appropriée.

Français

Numéro du catalogue
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Index

11

et la plage d’accord approriée

Utilisation de l’appareil
Symboles imprimés sur le brassard
SymbolesFonction/Description

Action recommandée
Réglez la marque ● sur l’artère dubras supérieur ou alignez-la surl'annulaire sur l'intérieur du bras.

Marque de position de l’artère

▲
REF

Index
Numéro du catalogue
Plage d’ajustement appropriée pour lebrassard adulte.
Imprimé sur le brassard adulte.
Utilisez le brassard adulte dePlage supérieure imprimée sur le brassard grande taille au lieu du brassard
adulte et sur le brassard de grande taile. adulte ou du brassard de grande
taile.
Plage d’ajustement appropriée pour lebrassard de grande taille.
Imprimé sur le brassard de grande taille.
Plage d’ajustement appropriée pour lebrassard adulte de grande taille.
Imprimé sur le brassard adulte de grande taille.
le brassard adulte de petite
Plage inférieure imprimée sur le brassard Utilisez
tailleaulieu du brassard adulte ou
adulte et sur le brassard de grande taille. du brassard
de grande taile.
Plage d’ajustement appropriée pour lebrassard adulte de petite taille.
Imprimé sur le brassard adulte de petite
taille.
Imprimé sur le brassard adulte de grande
taille et sur le brassard adulte de petite
taille.

Utilisez le brassard adulte au lieu
du brassard adulte de grande
taille ou du brassard adulte depetite taille.

Numéro de lot

Plage d’accord appropriée
Brassard adulte
de grande taille

A

L

Brassard de
grande taille

S

W

L

Brassard adulte

S

A

L

S

A

Brassard adulte
de petite taille

Français

12

Utilisation de l’appareil
Comment faire des mesures correctes

Pour une meilleure précision des mesures de la pression sanguine :Asseyez-vous confortablement sur une chaise. Posez le bras sur la table. Ne
croisez pas les jambes. Gardez les pieds au sol et tenez le dos bien droit.
Se relaxer pendant 5 à 10 minutes avant de commencer les mesures.
Lever le bras de sorte que le centre du brassard soit au même niveau que le cœur.
Bouger le moins possible et rester calme pendant les mesures.
Ne pas faire de mesure juste après un effort physique ou un bain. Se
reposer pendant 20 à 30 minutes avant de faire les mesures.
S’efforcer de mesurer la tension à la même heure tous les jours.

Pendant la mesure

Pendant la mesure, il est normal que l'on sente que le brassard est très
serré. (Ne pas s'inquiéter.)

Après la mesure

Après la mesure, appuyez sur le bouton START pour éteindre l’appareil.
Enlevez le brassard et notez votre tension et votre pouls. L'appareils’éteint automatiquement après une minute sans utilisation.

Français

13

Les mesures
Avant la mesure, voir «Consignes et observations pour une mesure correcte» à lapage 16.

Mesure normale
1.
2.

3.

Au niveau du cœurAsseyez-vous confortablement en mettant lebrassard au bras (gauche de préférance).
Appuyez sur le bouton START (mise en marche).
Appuyez
Tous les segments de l’affichage apparaissent.
Puis, zéro (0) s’affiche en clignotant un instant.
Tous les segments dePuis, l’affichage change comme indiqué dans la l’affichage apparaissent
brièvement.
figure de droite, lorsque la mesure commence. Le

brassard commence à gonfler. Il est normal que l’on
sente que le brassard serre fortement le bras. Un
indicateur de progression du gonflage est affiché,
dans l'angle gauche de la fenêtre, pendant legonflage.
Note: Si vous désirez arrêter le gonflage quelqueque soit le moment, appuyez de nouveau
sur le bouton START. (mise en marche).
Lorsque le brassard est entièrement gonflé, ilcommence à dégonfler automatiquement et lesymbole en forme de cœur () apparaît pour
indiquer que la mesure est en cours.
Une fois le pouls détecté, le symbole clignote àchaque pulsation.

START

Zéro s’affiche
Gonflage commence

Mise souspression

Note:

4.

5.

Si la pression correcte n'est pas obtenue,
l'appareil recommence le gonflage dubrassard automatiquement. Pour éviter unnouveau gonflage, voir la page suivante«Mesure avec la pression configurée».
Lorsque les mesures sont finies, les lectures
Mesuragedes pressions systoliques et diastoliques ainsi
en cours
que le nombre de pulsations sont affichées.
Le brassard finit de se dégonfler en chassantl'air restant.
Pression systolique
Appuyer sur le bouton START pour éteindre l’appareil.
L'appareil s’éteint automatiquement après une
minute sans utilisation.

Note: Attendez au moins trois minutes avant deremesurer votre tension (ou celle d’une
autre personne) à des fins de confirmation.

Français

14

Pression diastoliqueClassification suivant l’OMS

Pouls
L’air restant est chassé
automatiquement

Les mesures
Le Modèle UA-1020-W est étudié pour détecter le pouls et pour gonfler le brassard
jusqu’au niveau de la pression systolique automatiquement.
Si un nouveau gonflage se produit de manière répétée, utilisez les méthodes suivantes.

Mesure avec la pression configuréePendant la mesure de la pression sanguine, un nouveau gonflage peut survenir. Il est
possible de configurer une valeur de pression fixe afin d'éviter ce nouveau gonflage.
1.

2.

À l'étape 6 de la procédure d'ajustement del'horloge (reportez-vous à la page 10), appuyez surle bouton REGLAGES pour passer au mode deréglage de la pression. Le réglage actuel clignote.
ouAppuyez sur le boutonoupour sélectionnerune valeur de pression d'environ 30 mmHg, ou
plus, au-dessus de votre pression systoliqueprévue parmi les suivantes.
AUTO : Pressurisation automatique (valeur par défaut)
180 : Valeur de pression de 180 mmHg (fixe)
210 : Valeur de pression de 210 mmHg (fixe)
Éteindre
240 : Valeur de pression de 240 mmHg (fixe)

3.

Mode de réglage
TriCheckAppuyez sur le bouton REGLAGES pour passer au mode de réglage TriCheck.
Appuyez sur le bouton START pour éteindre l’appareil. L'appareil s’éteint
automatiquement après trois minutes sans utilisation.
La mesure suivante est réalisée avec la nouvelle valeur de pression.

Mesure avec la pression systolique désirée
Utilisez cette méthode si un nouveau gonflage se produit à plusieurs reprises,même si la valeur de pression est réglée sur 240 dans la procédure ci-dessus ou si
les résultats ne s'affichent pas même si la pression diminue de 20 mmHg ou moins.
1.

Mettez le brassard au bras (gauche depréférence).

Au niveau du cœur2.

Appuyez et maintenez enfoncé le bouton
START jusqu’à ce qu’un nombre 30 à 40
mmHg plus grand que votre pression
systolique prévisible n’apparaisse.

Appuyez et maintenez
enfoncé le bouton

3.

Relâcher le bouton
START pourcommencer la mesure, quand le nombre
désiré est indiqué. Ensuite continuez àmesurer votre tension artérielle commedécrit à la page précédente.
Français

15

START

Relâcher le bouton
quand la pression
systolique désirée est
obtenue

Se référer à la page précédente
pour la mesure.

Consignes et observations pour une mesure correcte
Asseyez-vous confortablement. Posez le bras sur une table, la paume
dirigée vers le haut et le brassard au même niveau que le cœur.
Détendez-vous pendant cinq à dix minutes avant de commencer lamesure. Si vous êtes énervé ou au contraire déprimé, cela se reflètera
sur votre tension de même que sur votre pouls.
La tension de tout individu varie constamment selon ce qu’il fait et ce qu’ilmange. D’autre part, certaines boissons ont un effet très fort et très
rapide sur la tension artérielle.
Cet appareil se réfère aux pulsations cardiaques pour mesurer la tension
et le pouls. Par conséquent, si vos pulsations sont faibles ou irrégulières,l’appareil pourrait avoir des difficultés à mesurer votre tension artérielle.
Au cas où l’appareil détecte une condition anormale, il s’arrête demesurer et affiche un symbole d’erreur. Voir la page 6 pour la description
des symboles.
Cet appareil est conçu pour être utilisé par des adultes. Avant del’utiliser pour mesurer la tension d’un enfant, veuillez en discuter avec
votre médecin. Ne laissez jamais un enfant prendre lui-même sa tension
sans surveillance.
La performance du tensiomètre automatique peut être affectée par unetempérature, une humidité ou une altitude excessives.

Mesure TriCheckTM
Le mode TriCheckTM prend automatiquement trois mesures consécutives
et affiche les valeurs moyennes de celles-ci.
Dans le mode de réglage de lapression

Sélection du mode TriCheckTM
1.

Dans le mode de réglage de la pression,
appuyez sur le bouton REGLAGES pouraccéder au mode de réglage TriCheckTM.
Le réglage actuel clignote.

2.

ou
pour
Appuyez sur le bouton
activer ou désactiver le modeLe réglage actuel clignote.
TriCheckTM.
TM
ON: Mode TriCheck activé
ou
OFF: Mode de mesure normal (valeur par défaut)3.

Appuyez

Modifier le réglage

Appuyez sur le bouton START ou
REGLAGES pour éteindre l’appareil.
L'appareil s’éteint automatiquement
après trois minutes sans utilisation.

Français

ou

16

START

Mesure à l'aide du mode TriCheckTM
1.

2.

Appuyez sur le bouton START
Tous les segments de l’affichage
apparaissent.
Puis, zéro (0) s’affiche en clignotant
un instant.
Puis, la première mesure démarre.
Lorsque les mesures sont finies, leslectures des pressions systoliques etdiastoliques ainsi que le nombre depulsations sont affichées.
Puis un compte à rebours d'une minutedémarre pour la seconde mesure.

3.

Après une minute, la seconde mesuredémarre.

4.

Lorsque les mesures sont finies, leslectures moyennes de la première et dela seconde mesure s'affichent.
Puis un compte à rebours d'une minutedémarre pour la troisième mesure.

5.
6.

Au niveau du cœurSTART

Appuyez
Tous les segments del’affichage apparaissent
brièvement

Première mesure

Une minute plus tard

Après une minute, la troisième mesuredémarre.

Résultats de lapremière mesure

Seconde mesure

Lorsque les mesures sont finies, leslectures moyennes des trois mesuress'affichent et sont stockées en mémoire.

Note
Le symbole TriCheckTM s'affiche pendant
la mesure.
Pour annuler la mesure, appuyez sur lebouton START. Dans ce cas, aucune
donnée n'est stockée en mémoire.
Si le symbole d'erreur de fixation dubrassard s'affiche pendant la première
mesure, annulez-la, appliquez de nouveau
le brassard de manière appropriée etdémarrez une nouvelle mesure.
Après la mesure, les données moyennes
des trois mesures sont stockées enmémoire.
Aucune donnée n'est stockée, lors del'annulation
desmesures,avant
l'achèvement de la troisième mesure.
Français

Une minute plus tard

Moyenne des première
et seconde mesures

Troisième mesure

Moyenne des trois
mesures17

Rappel des données mémoriséesNote: Cet appareil mémorise les 90 dernières mesures.
Appuyezou

1. Appuyez sur le boutonou .
La moyenne des données de toutes lesmesures et le nombre de données sontaffichés.
(S'il n'y a pas de données, « 0 » s'affiche.
ouAppuyez sur le bouton ,START pour éteindre l’appareil.)Pression systolique moyenne
Pression diastolique moyenne
Pouls moyen

Moyenne de toutes
les données

2. Ensuite, chaque pression sur le bouton(ou le bouton
pour afficher lesdonnées dans l'ordre inverse), affiche
les données mémorisées comme suit.

Pression systolique moyenne
Pression diastolique moyenne

Moyenne de toutes les mesures AM (dumatin) prises entre 4h00 et 9h59.
(Dans l’exemple, 10 mesures. S'il n'y apas de données, « -- » s'affiche.)Pouls moyen
Moyenne desdonnées AM

Moyenne de toutes les mesures PM (dusoir) prises entre 18h00 et 1h59.
(Dans l’exemple, 9 mesures. S'il n'y a pasde données, « -- » s'affiche.)Pression systolique moyenne
Pression diastolique moyenne
Pouls moyen

Données les plus récentes (N° 35 dans

Moyenne desdonnées PM

l’exemple)
Trois secondes après que les chiffres des donnéessont affichés, les données mesurées s’affichent.
Dernières données (N° 1)
Trois secondes après que les chiffres des donnéessont affichés, les données mesurées s’affichent.
3. Une fois les dernières données affichées,appuyez sur le bouton pour revenir àl’affichage de la moyenne de toutes les mesures.
4. Appuyez sur le bouton START pour éteindre
l’appareil. L'appareil s’éteint automatiquement
après une minute sans utilisation.
Français

PressionsystoliquePression
diastolique
PoulsDonnées les plus récentes

PressionsystoliquePression
diastolique
PoulsDernières données (les plus anciennes)

18

Qu’est ce que l’indicateur de RCI/FA?
Lorsque le moniteur détecte un rythme irrégulier pendant les mesures,l’indicateur de RCI/FA apparaît à l’écran avec les valeurs mesurées.
Note: Nous vous recommandons de contacter votre médecin si vous voyez
fréquemment cet indicateur de RCI/FA :.

Qu’est-ce que la FA?
Le cœur se contracte en raison des signaux électriques se produisant dans lecœur et envoie le sang dans tout le corps. La fibrillation auriculaire (FA) se
produit lorsque le signal électrique dans l’oreillette est troublé et provoque desperturbations dans l’intervalle des pulsations. La FA peut causer la stagnation
du sang dans le cœur, ce qui peut facilement créer des caillots sanguins, cause
d’AVC et de crise cardiaque.

Indicateur de progression du gonflage
L’indicateur montre la progression de la pression pendant la mesure.
Mise sous pression

Dégagement d’air

Gonflage en cours

Mesure en cours
Gonflement complet

Indicateur de tension selon la classification OMS
Chaque segment d’indicateur de progression
de la pression correspond à la classification dela pression sanguine selon l’OMS décrite à lapage suivante.
Indicateur de la classification selon l’OMS

Hypertention sévère
Hypertention modérée
Hypertention faible
Haute-Normale
NormaleOptimale

Exemple:Hypertention modérée

Hypertention faible: L’indicateur affiche un segment, basé surles dernières données, correspondant à laclassification suivant l’OMS.
Haute-Normale
Français

19

Pression sanguine
Qu'est-ce que la pression sanguine?
La pression sanguine est la force avec laquelle le sang vient frapper contre les parois
des artères. La pression systolique se produit lorsque le cœur se contracte. La
pression diastolique se produit lorsque le cœur se détend.
La pression sanguine est mesurée en millimètre de mercure (mmHg). La pression
sanguine naturelle d’un individu est représentée par la pression fondamentale qui est
mesurée avant toute activité, le matin au lever du lit, et avant de manger ou de boire.

Qu'est-ce que l'hypertension et comment la contrôler?
L’hypertension est une tension artérielle (ou plus exactement une «pression
sanguine») anormalement élevée. Non surveillée, l’hypertension peut provoquer denombreux problèmes de santé, notamment une attaque d’apoplexie ou une crise
cardiaque.
L'hypertension peut être contrôlée en changeant son style de vie, en évitant tout
stress et avec des médicaments pris sous surveillance médicale.
Pour empêcher l’hypertension ou la garder sous contrôle:
Ne pas fumer.
Faire régulièrement des exercices
Réduire sa consommation de sel etphysiques.
de graisse.
Effectuer régulièrement desMaintenir un poids adéquat.
visites médicales.

Comment mesurer la pression sanguine chez soi?
Lorsque la tension est mesurée à l’hôpital ou dans le cabinet du médecin, on neparvient pas toujours à se décontracter, ce qui peut avoir pour effet d’augmenter latension de 25 à 30 mmHg par rapport à la tension mesurée chez soi. Le fait demesurer sa tension tranquillement chez soi réduit les influences extérieures sur lavaleur de la tension, complète les mesures effectuées par le médecin et permetd’établir un relevé plus précis et plus complet de sa tension artérielle.

Classification de la tension selon l’OMS
L’évaluationdel’hypertension,
indépendamment du critère âge, a été établie
par l’Organisation Mondiale de la Santé(OMS) selon le tableau ci-joint.

Variations de la pression sanguine
La tension artérielle (ou pression sanguine)d’un individu varie fortement selon le momentde la journée et selon la saison. Ainsi, ellepeut varier de 30 à 50 mmHg dans une mêmejournée selon un ensemble de conditions.
Chez les personnes hypertendues, cesvariations sont souvent encore plus prononcées.
Normalement, la pression augmente pendant le travail ou le jeu et tombe à son plus
bas niveau pendant le sommeil. Par conséquent ne pas trop s'inquiéter des résultats
d'une seule mesure.

Français

20

Mesurez votre tension tous les jours à lamême heure (en suivant la procédure décritedans ce manuel) pour connaître votre
pression sanguine normale. Ces mesuresquotidiennes vous permettront d’établir unrelevé plus complet de votre tension.
Veillez à bien noter la date et l'heure lorsque
vous enregistrez votre pression. Consultez
votre médecin pour interpréter les données
recueillies.

Résolution des problèmes
Problème
Rien n'apparaît sur

Cause probable

Solution recommandéeLes piles sont usées.

l'affichage, mêmelorsque l'on met
l'appareil en marche.

Les bornes des piles ne sontpas dans le bon sens.

Remplacer les piles usées par desneuves.
Mettre les piles dans le bon sens en
faisant correspondre la borne positiveet négative comme indiqué sur lebrassard.

La tension des piles est trop
Le brassard ne segonfle pas.

faible.
Le symbole piles faibles
apparaît. Si les piles sont usées,

Remplacer les piles usées par desneuves.

ce symbole n’apparaît pas.
L’appareil ne fait pas
les mesures. Leslectures sont trop

Le brassard est mal serré.
Vous avez bougé votre bras ou
autre pendant la mesure.
Le brassard est mal placé.

hautes ou trop
basses.

Autres

La valeur est différente de cellemesurée dans une clinique ou
chez le médecin.

Serrer correctement le brassard.
Ne pas bouger et rester calmependant la mesure.
S'asseoir confortablement et ne pasbouger. Posez le bras sur une table, lapaume dirigée vers le haut et le brassard
au même niveau que le cœur.
Si votre pouls est faible ou irrégulier,l'appareil aura des difficultés àdéterminer votre pression sanguine.
Voir le paragraphe "Comment
mesurer la pression sanguine chezsoi ?".
Retirer les piles. Le remettre
correctement en place et essayer à
nouveau de mesurer.

Note: Si les solutions décrites ci-dessus ne suffisent pas pour résoudre les problèmes,contactez votre revendeur. N’essayez pas d’ouvrir ou réparer vous-même cetappareil car toute tentative invaliderait votre garantie

Français

21

Entretien
Ne pas ouvrir l'appareil car il comporte des composants électriques sensibles et
une unité d'air complexe qui pourraient être endommagés. Si vous n'arrivez pasà situer un problème avec le tableau de recherche de problèmes, demandez auservice après vente de votre revendeur ou du groupe de service A & D. Le
groupe de service A & D vous fournira des informations techniques sur lesfournisseurs agréés, les pièces et les unités de rechange.
Cet appareil a été conçu et fabriqué pour durer. Il est généralement
recommandé de le faire vérifier tous les 2 ans pour s’assurer de son
fonctionnement correct et de sa précision. Veuillez contacter votre revendeur
agréé A&D pour l’entretien.

Fiche techniqueType

UA-1020-W

Méthode de mesure

Mesure oscillométrique

Plage de mesure

Pression : de 0 à 299 mmHg
Pression systolique : de 60 à 279 mmHg
Pression diastolique : de 40 à 200 mmHg
Pouls :de 40 à 180 pulsations / minute

Précision de mesure

Pression : ±3 mmHg
Pouls : ±5 %Alimentation électrique

4 x 1,5 piles (R6P, LR6 ou AA) ou
adaptateur en courant alternatif (TB-233C) (Nonfournis)Nombre de mesures

Env.1000 fois LR6 (piles alcalines)
Env. 300 fois R6P (piles au manganèse)
Avec une valeur de tension de 180 mmHg à une
température ambiante de 23 °C

Classification

Équipement a alimentation interne ME (fournie parles piles)
Class II (fournie par l'adaptateur)Mode de fonctionnement en continu

Test clinique

Conformément à ISO81060-2：2013
Dans l’étude de validation clinique, K5 a été utilisé
sur 85 sujets pour la détermination de la tension
artérielle diastolique.

EMD

IEC 60601-1-2: 2014

Mémoire de données

Dernières 90 mesures

Condition de fonctionnement

de +10 à + 40 °C (hum. rel. de 15 à 85 %)de 800 à 1060 hPa

Français22

Condition de stockage / transport
Dimensionsde -20 à + 60 °C (hum. rel. de 10 à 95 %)de 700 à 1060 hPa
environ 140 [l] x 60 [h] x 105 [p] mmProtection

Appareil: IP20

Poids

environ 285 g sans les piles

Partie appliquée

Brassard Type BF

Durée de vie utile

Appareil: 5 ans (six utilisations par jour)Brassard: 2 ans (six utilisations par jour))
Adapteur secteur : 5 ans (lorsqu'utilisé six fois par jour)

Accessoire :Adaptateur secteur TB-233C.

L’adaptateur est utilisé pour connecter la source depuissance chez soi.
Veuillez prendre contact avec votre revendeur A&D
local pour l’achat de cet accessoire.
L’adaptateur secteur doit être inspecté ou remplacé
périodiquement.
Symboles imprimés sur l’adaptateur secteur
SymbolesFonction / Signification
Utilisation à intérieur uniquement
Appareil de classe II
Fusible thermique
FusibleLabel de conformité de l’appareil à ladirective CE
Label de conformité de l’appareil à lacertification EAC
Polarité de la fiche de l’adaptateur secteur

Accessoires vendus séparément
Brassard
Numéro du catalogue
Taille du brassard
CUF-F-LA
Brassard adulte de grande taille
CUF-I
Brassard de grande taille
CUF-F-A
Brassard adulte
CUF-F-SA
Brassard adulte de petite taille
Adaptateur secteur
Numéro du catalogue
TB-233C

Taille du bras
31 cm à 45 cm
22 cm à 42 cm
22 cm à 32 cm
16 cm à 24 cm

Fiche

Type C

Note: Ces caractéristiques techniques sont sujettes à changement sans préavis.
Le classement IP indique les degrés de protection fournis par les coffrets deprotection conformément à la norme CEI 60529. Cet appareil est protégé
contre les corps étrangers solides de 12 mm de diamètre et plus, tel qu’undoigt. Cet appareil n’est pas protégé contre l’eau.

Français

23

MEMO

Français24ÍNDICE
Estimados clientes ................................................................................................. 2
Observaciones preliminares .................................................................................. 2
Precauciones .......................................................................................................... 2
Partes de identificación de componentes................................................................. 4
Símbolos ................................................................................................................. 5
Modo de operación ................................................................................................ 7
Uso del monitor ...................................................................................................... 8
Instalación/Cambio de las pilas........................................ 8Conexión de la manguera de aire ..................................... 9Conexión del adaptador de CA ......................................... 9
Ajuste del reloj incorporado ........................................... 10
Selección del brazalete correcto ..................................... 11
Cómo colocar el brazalete .............................................. 11
Para efectuar mediciones correctas ................................ 13
Medición ........................................................................ 13
Después de la medición ................................................. 13
Mediciones............................................................................................................ 14
Medición normal ............................................................ 14
Medición con la presión ajustada ................................... 15
Medición con la presión sistólica deseada ...................... 15
Notas para mediciones correctas..................................... 16
Medición TriCheckTM ............................................................................................ 16
Revisión de los datos de la memoria .................................................................. 18¿ Qué es el indicador de LI/FibA? ....................................................................... 19¿ A qué hace referencia FibA?............................................................................ 19
Indicador de barra de presión.............................................................................. 19
Indicador de clasificación de la OMS .................................................................. 19
Aspectos sobre la tensión arterial ....................................................................... 20¿Qué es la tensión arterial?............................................. 20¿Qué es la hipertensión y cómo controlarla? ................... 20¿Por qué debe medirse la tensión arterial en la casa? ...... 20
Clasificación de tensión arterial según la OMS ................ 20
Variaciones de tensión arterial ....................................... 20
Localización y corrección de fallos ...................................................................... 21
Mantenimiento ...................................................................................................... 22
Datos técnicos ...................................................................................................... 22

Español

1

Estimados clientes
FELICIDADES. Usted ha comprado un monitor de presiones sanguíneas
A&D, el aparato tecnológico más avanzado y más fácil de usar
actualmente en el mercado.
Se recomienda leer este manual de instrucciones detenidamente
antes de usarlo por primera vez.

Observaciones preliminares
Este aparato se ajusta a la Directiva Europea 93/42 EEC para productos
médicos. Esto se evidencia con la marcade conformidad.
(0123: Número de referencia notificado al cuerpo interesado)
Este aparato está diseñado para usarse en adultos, los niños y los reciénnacidos no pueden usarlo.
Ámbito de uso. Este aparato es para que pueda usarlo usted mismo en
un entorno doméstico.
Este aparato se ha diseñado para medir la presión sanguínea y lafrecuencia del pulso de las personas para el diagnóstico.

Precauciones
En la construcción de este aparato, se han usado componentes deprecisión. Deberán evitarse extremos de temperatura, humedad, rayos
solares directos, golpes o polvo.
Limpie el aparato y el brazalete con un paño seco y suave humedecido con
agua y jabón neutro. No utilice nunca alcohol, benceno, diluyentes u otros
productos químicos agresivos para limpiar el aparato o el brazalete.
Evite doblar el brazalete apretado o guardar el tubo retorcido durante períodos
prolongados, dado que ello puede acortar la vida útil de los componentes.
Tenga cuidado para evitar la estrangulación accidental de niños o bebéscon el tubo o el cable.
No retuerza la manguera de aire durante la medición. Si lo hace, pueden
producirse lesiones debido a la presión continua del brazalete.
El aparato y el brazalete no son resistentes al agua. Por lo tanto, evite
que la lluvia, la transpiración y el agua mojen el aparato y el brazalete.
Il est possible que les mesures soient altérées si l’appareil est utilisé à
proximité de téléviseurs, de fours à micro-ondes, de téléphones
cellulaires, de sources de rayonnement X ou d’autres appareils émettant
de forts champs électriques.
Los aparatos de comunicación inalámbrica, como los dispositivos de red doméstica,
teléfonos móviles, teléfonos inalámbricos y sus centrales y walkie-talkies, pueden
afectar al monitor de presiones sanguíneas. Por tanto, debe mantenerse una
distancia mínima de 30 cm respecto a dichos aparatos.
El aparato, sus partes y pilas no deben considerarse como desperdicios domésticos
ordinarios y deberán eliminarse de acuerdo con los reglamentos aplicables.
Cuando se usa el adaptador de CA, asegúrese de que se puede retirarrápidamente el adaptador de CA del enchufe eléctrico cuando sea necesario.
Español2

Cuando se reutiliza el aparato, confirme que el aparato esté limpio.
No modifique el aparato. Puede provocar accidentes o daños al aparato.
Para medir la presión sanguínea, el brazo debe ser apretado por el brazalete
lo suficiente como para detener temporalmente el flujo sanguíneo a través dela arteria. Esto puede provocar dolor, entumecimiento o una marca roja
temporal en el brazo. Este estado aparecerá especialmente cuando la
medición se repite sucesivamente.
Cualquier dolor, entumecimiento o marcas rojas desaparecerán con el tiempo.
Si la presión sanguínea se mide con demasiada frecuencia, pueden
producirse daños debido a la interferencia con el flujo sanguíneo. Si utiliza elaparato de forma repetida, compruebe que su uso no provoca una alteración
prolongada de la circulación sanguínea.
Si le han practicado una mastectomía, consulte con su médico antes deutilizar el aparato.
No deje que los niños utilicen el dispositivo sin la supervisión de un adulto, ni
utilice el dispositivo en un lugar dentro del alcance de los niños. Podría causar
accidentes o daños.
El aparato contiene piezas pequeñas que pueden provocar un
atragantamiento si los niños las tragan accidentalmente.
Desconecte el adaptador de CA cuando no se use durante la medición.
El uso de accesorios no indicados en este manual puede afectar a laseguridad.
Si se produce un cortocircuito en la pila, puede calentarse y posiblementecausar quemaduras.
Permita que el aparato se ajuste al entorno circundante antes de usarlo(alrededor de una hora).
No se han realizado estudios clínicos en bebés recién nacidos o mujeres
embarazadas. No lo use en niños recién nacidos o mujeres embarazadas.
No toque las pilas, la clavija de CC y al paciente simultáneamente. Podría
ocasionar una descarga eléctrica.
No lo hinche sin colocar el brazalete en la parte superior del brazo.

Contraindicaciones

A continuación se encuentran precauciones para el uso correcto del aparato.
No aplique el brazalete a un brazo con otro equipo eléctrico médico
acoplado. Es posible que el equipo no funcione correctamente.
Las personas que tienen una deficiencia circulatoria grave en el brazo
deben consultar a un médico antes de usar el aparato, para evitar
problemas médicos.
No autodiagnostique los resultados de la medición ni se ponga un
tratamiento usted mismo. Consulte siempre a su médico para laevaluación de los resultados y el tratamiento.
No aplique el brazalete a un brazo con una herida no curada.
No aplique el brazalete a un brazo que está recibiendo un goteo
intravenoso o transfusión sanguínea. Puede provocar lesión oaccidentes.
No utilizar el aparato cuando se encuentran presentes gases inflamables
como, por ejemplo, gases anestésicos. Puede provocar una explosión.
No usar el aparato en entornos con alta concentración de oxígeno, comopor ejemplo en, cámaras de oxígeno a alta presión o una tienda deoxígeno. Puede provocar un incendio o explosión.
Español

3Partes de identificación de componentes
Jack de CC

Pantalla

Botón
Botón

Tapón del conectorde aireBotón AJUSTAR
Botón START

Zócalo de aire

Manguera deaire
Marca de posición
de la arteria

Rango de colocación

Brazalete
Marca índice
Cubierta de pilas
Compartimiento de pilas

Pilas de 1,5 V(R6P, LR6 o AA)

Indicadores dela pantalla

Promedio

Memoria

Ajustes de presión

Símbolo LI/FibA

Símbolo de error de ajuste del brazalete
Presión
sistólica

Símbolo de error por movimiento
Símbolo de TriCheckTM

Presión
diastólica

Marca de corazón
Indicador de clasificación de OMS e
indicador de la barra de presión
Indicador de pilas

Frecuencia delpulso

Marca de AM / PM
Español

Display del reloj
4

Símbolos
Símbolos impresos en la carcasa del aparato y en el adaptador de CA
SímbolosSN

Función / Significado

Acción recomendada

Aparato en “standby” y botón de encendido
Guía para instalación de pilas
Corriente continua
Número de serie
Fecha de fabricación
Tipo: BF; El aparato, el brazalete y tubo
están diseñados para proporcionar
protección especial contra
electrochoques.
Etiqueta de aparatos médicos según
directiva de CE
Etiqueta de WEEE
Fabricante
Representante UE
Consulte el manual/folleto de instrucciones

IP

Polaridad del jack de CC
Símbolo de protección internacional
Mantener seco

Símbolos que aparecen en la pantalla
SímbolosFunción / Significado

Acción recomendada

Aparece cuando la medición está en curso.
Este indicador parpadea cuando se detecta
el pulso.

La medición está en
curso.
Quédese muy tranquilo
durante la medición.

Símbolo LI/FibA
Este indicador aparece cuando se detectanlatidos irregulares del corazón. Puede
iluminarse cuando se detecta una vibración
muy ligera, como un escalofrío o un temblor.
El modo TriCheckTM realiza tres mediciones
consecutivas de forma automática ymuestra los valores promedio de las tres
mediciones.
Símbolo que aparece cuando se
detecta un movimiento del cuerpo o elbrazo.
Español

5

La lectura puede ofrecer
un valor incorrecto.
Vuelva a medir otra vez.
Quédese muy tranquilo
durante la medición.

Símbolos

Función / Significado

Acción recomendada

Símbolo que aparece durante la
medición cuando el brazalete no estábien ajustado.

La lectura puede ofrecer
un valor incorrecto.
Coloque el brazaletecorrectamente, y vuelva amedir otra vez.

Las mediciones previas almacenadas
en la memoria.
Datos medios
BATERÍA CARGADA

Indicador de la potencia de las pilas
durante la medición.
Reemplace todas las pilas
Las pilas son débiles cuando parpadea con unas nuevas, cuando
destelle esta marca.
esta marca.
Vuelva a medir otra vez.
Presión sanguínea inestable debido al
Quédese muy tranquilo
movimiento durante la medición.
durante la medición.
Los valores sistólico y diastólico seencuentran dentro de 10 mmHg.
El valor de presión no ha aumentado
durante la inflación.
Coloque el brazalete
BATERÍA BAJA

correctamente, y vuelva amedir otra vez.

El brazalete no está colocado
correctamente.
ERROR DE VISUALIZACIÓN DEL IMPULSOEl impulso no se ha detectado
correctamente.

Error interno del monitor de presiones
sanguíneas

SYS
DIA
PUL
AM
PM

Presión sanguínea sistólica indicada en mmHg.
Presión sanguínea diastólica indicada en mmHg.
Frecuencia del pulso por minuto
Datos tomados entre las 4:00 y las 9:59 horas
Datos tomados entre las 18:00 y las 1:59 horas
Ajustes de presión
Indica el valor de presión previamente
ajustado por el usuario.
Español

6

Extraiga las pilas, presione elbotón START y a continuación
vuelva a colocar las pilas. Si elerror sigue apareciendo,
póngase en contacto con su
distribuidor.

Modo de operación
1. Medición normal con los datos almacenados
Presione el botón START. Se mide la presión sanguínea y sus datos sealmacenan en el aparato. El aparato puede almacenar los últimos 90 datos
automáticamente en la memoria.

2. Revisión de los datos de la memoria
Presione el botóno para revisar los datos.
Aparece el promedio de datos de todas las
mediciones, tal y como se indica en la figura de
la derecha.
A continuación, cada vez que se presione elbotón , se mostrarán los datos de la memoriade la siguiente forma.
Promedio de todas las mediciones AM (mañana)realizadas entre las 4:00 y las 9:59 horas.
Promedio de todas las mediciones PM(tarde/noche) realizadas entre las 18:00 y las 1:59
horas.
Datos más recientes (No. n, en el ejemplo, No. 35)
Último dato (No.1)
Para más detalles acerca de la revisión de datos,
consulte “Revisión de los datos de la memoria”.
(Véase página 18.)

3. Borrado de datos los datos almacenados en la memoria
Presione los dos botones
y . Aparecerán la marcayel indicador de pilas. Mantenga presionados los dos botonesyhasta que la marca
iluminada empiece a parpadear
para borrar todos los datos almacenados en la memoria.

4. Medición con la presión sistólica deseada
Véase página 15 para la medición con la presión sistólica deseada.

Español7

Uso del monitor
Instalación/Cambio de las pilas

Paso 1

1. Retire la cubierta de pilas.

Paso 3

2. Retire las pilas usadas e inserte lasnuevas en el compartimiento tal como se
describe, respetando las polaridades (+ y -)de las pilas.
Utilice solo las pilas R6P, LR6 o AA.

Paso 1

Paso 2

3. Cierre la cubierta de pilas.

PRECAUCIÓN
Coloque las pilas en el compartimiento de las pilas, tal como se
muestra en la figura. De lo contrario, el aparato no funcionará.
Cuando la marca(marca de PILAS GASTADA) parpadea en la pantalla, reemplace
todas las pilas por las nuevas. No mezcle pilas viejas con las nuevas, dado queesto puede acortar la vida útil de las mismas o causar un mal malfuncionamientodel aparato.
Sustituya las pilas al menos dos segundos después de que se apague el aparato.
(marca de PILAS GASTADA) aparece incluso después de sustituir las pilas,
Si la marca
realice una medición de la presión sanguínea. El aparato podría entonces
reconocer las pilas nuevas.
La marca(marca de PILAS GASTADA) no aparece cuando las pilas están totalmenteagotadas.
La vida útil de las pilas varía con la temperatura ambiente y puede acortarse atemperaturas bajas. Por lo general, cuatro pilas LR6 nuevas duran
aproximadamente un año si se usan para realizar mediciones dos veces al día.
Use solamente las baterías especificadas. Las baterías suministradas con el aparato
son para probar el funcionamiento del monitor y pueden tener una vida útil más corta.
Retire las pilas cuando no vaya a usar el aparato durante un período prolongado.
Las pilas pueden tener fugas y causar un malfuncionamiento.

Español8

Uso del monitor
Conexión de la manguera de aire

Zócalo de aire

Inserte firmemente el tapón delconector de aire en el zócalo deaire.

Tapón del conector de aire

Conexión del adaptador de CA
Inserteelenchufedeladaptador de CA en el jack de
CC. A continuación, inserte eladaptador de CA en una tomade corriente.

Enchufe adaptador de CA

Utilice el adaptador de CA
especificado.
(Véase página 23.)

Jack de CC

Cuando desconecte el adaptador de CA de la toma de corriente, agarre
el armazón del adaptador de CA y tire de él para extraerlo de la toma decorriente.
Cuando desconecte el enchufe del adaptador de CA del monitor depresiones sanguíneas, agarre el enchufe del adaptador de CA y tire deél para extraerlo del monitor.

Español9

Uso del monitor
Presione

Ajuste del reloj incorporado
Ajuste el reloj antes de usarlo.

oAño

1. Presione el botón AJUSTAR hasta que losdígitos comiencen a destellar.
2. Seleccione el año utilizando el botóno .
Presione el botón AJUSTAR para ajustar el año
en curso y pase a la selección del mes/día. El
año puede ajustarse entre el 2010 y el 2059.
3. Seleccione el mes utilizando el botóno .
Presione el botón AJUSTAR para ajustar el mes
en curso y pase a la selección del día.
4. Seleccione el día utilizando el botóno .
Presione el botón AJUSTAR para ajustar el día en
curso y pase a la selección de la hora /minuto.
5. Seleccione la hora utilizando el botóno .
Presione el botón AJUSTAR para ajustar lahora en curso y pase a la selección del minuto.
6. Seleccione el minuto utilizando el botóno .
(Presione el botón AJUSTAR para ir al modo deajuste de presión. Para obtener más información,
consulte la página 15.)Presione el botón START para apagar el aparato.

o
Mes

Día

o

o

Hora

o
MinutoNota: Después de tres minutos de inactividad, el
aparato se apaga automáticamente.
Si no se ha ajustado el reloj, en lapantalla del reloj aparecen rayas
tal como se muestra a la derecha.
Si se mantiene presionado el botón, el valor cambia de formaoininterrumpida.

Español

ApagarEl aparato puedeapagarse en cualquier
momento presionando
el botón START.

10

Modo de ajustede presión

Uso del monitor
Selección del brazalete correcto
La selección de un brazalete de tamaño correcto es importante para la lectura
correcta. Si el brazalete no es de tamaño correcto, la lectura puede indicar un
valor incorrecto de la presión sanguínea.
El tamaño del brazo está impreso en cada brazalete.
El índicey el gama adecuada de colocación del brazalete le indicarán, si elbrazalete está colocado correctamente. (Refiérase a "Símbolos impresos en el
brazalete" en la página siguiente.)
Si el índice
se encuentra fuera de la gama, póngase en contacto con el
distribuidor local para adquirir el brazalete de reemplazo.
El brazalete es un producto consumible. Si se deteriora, adquiera uno nuevo.
Tamaño del brazo Tamaño del brazalete recomendado
31 cm a 45 cm
Brazalete grande de adulto
22 cm a 42 cm
Brazalete de amplio rango
22 cm a 32 cm
Brazalete de adulto
16 cm a 24 cm
Brazalete pequeño de adulto
Tamaño del brazo: La circunferencia en el bíceps.

Cómo colocar el brazalete

Evite que la manga quede demasiado
apretada al remangarse.

1. Coloque el brazalete alrededor del brazo,
unos 1-2 cm por encima de la parteinterior del codo, tal como se muestra en
la figura. Coloque el brazalete en contacto
directo con la piel, dado que la ropa puedecausar un pulso débil y provocar error de
medición.

Brazalet
1-2 cmManguera de aire
Marca deposición de la arteria

2. Asimismo, la constricción del brazo,
causada al remangarse la manga
apretada, no permitirá lecturas correctas.

Abroche bien elcierre de velcro.

se encuentre
3. Confirme que el índicedentro de la gama de buen estado.

ÍndiceEspañolNº de catálogo
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

11

y la rango de colocación

Uso del monitor
Símbolos impresos en el brazaleteSímbolosFunción/Significado

Acción recomendada
Ajuste la marca ● en la arteria delbrazo o de forma alineada con el dedo
anular en la parte interior del brazo.

●

Marca de posición de la arteria▲
REFÍndiceNúmero de catálogo
Rango de colocación adecuado parael brazalete de adulto.
Está impreso en el brazalete de adulto.
Por encima del rango impreso en el
brazalete de adulto y el brazalete deamplio rango.
Rango de colocación adecuado para el
brazalete de amplio rango.
Está impreso en el brazalete de amplio
rango.
Rango de colocación adecuado parael brazalete grande de adulto.
Está impreso en el brazalete grandede adulto.
Por debajo del rango impreso en el
brazalete de adulto y el brazalete deamplio rango.
Rango de colocación adecuado parael brazalete pequeño de adulto.
Está impreso en el brazalete pequeñode adulto.
Está impreso en el brazalete grandede adulto y el brazalete pequeño deadulto.

Utilice el brazalete grande de adulto
en lugar del brazalete de adulto obrazalete de amplio rango.

Utilice el brazalete pequeño deadulto en lugar del brazalete deadulto o brazalete de amplio rango.

Utilice el brazalete de adulto en
lugar del brazalete grande deadulto o brazalete pequeño deadulto.

Número de loteRango de colocación
Brazalete grande deadultoA

L

Brazalete de amplio
rango

S

W

L

Brazalete de adulto

S

A

L

S

A

Brazalete pequeñode adulto

Español

12

Uso del monitor
Para efectuar mediciones correctas
Tenga en cuenta lo siguiente para una óptima precisión en la medición de la tensión arterial:
Siéntese cómodamente en una silla. Apoye su brazo sobre la mesa. No cruce
las piernas. Mantenga los pies en el suelo y enderece la espalda.
Relájese durante unos 5 a 10 minutos.
Coloque el centro del brazalete al mismo nivel que su corazón
Permanezca quieto y en silencio durante la toma de la presión.
No efectúe la medición inmediatamente después de un ejercicio físico o baño.
Descanse unos 20 ó 30 minutos antes de la medición.
Trate de medir su tensión arterial a la misma hora todos los días.

Medición
Durante la medición, es normal que se sienta el brazalete muy apretado (no se
alarme por esto).

Después de la medición
Después de la medición, presione el botón START para apagar el aparato.
Retire el brazalete y registre sus datos de medición. Después de un minuto de
inactividad, el aparato se apaga automáticamente.

Español

13

Mediciones
Antes de la medición, lea “Notas para mediciones correctas” en la página 16.

Medición normal
1.

2.

Coloque el brazalete en su brazo (brazo
izquierdo preferiblemente).
Siéntese tranquilamente durante la medición.
Presione el botón START.
Aparecen brevemente todas las secciones dela pantalla. A continuación, parece cero (0)parpadeando brevemente. Luego la indicaciónde la pantalla cambia como se muestra en la figura
derecha, al comenzar la medición. El brazalete
comienza a inflarse. Es normal que el brazaleteapriete fuertemente. Se visualiza el indicador debarra de presión durante la inflación en el borde
izquierdo de la pantalla.
Nota:

3.

Al término de la inflación del brazalete, ladeflación comienza automáticamente y lamarca(marca de corazón) parpadea
indicando que la medición está en curso. Unavez detectado el pulso, esta marca parpadeará
coincidiendo con el latido del corazón.
Nota:

4.

5.

Si desea detener el inflado en cualquier
momento, presione el botón START
nuevamente.

A nivel del corazón

Presione

START

Aparecen brevemente
todas las secciones
de la pantalla.

Aparece ceroComienza la inflación

Presionamiento

Si no se obtiene una tensión apropiada, el
aparato empieza nuevamente el inflado
automáticamente.
Para evitar la reinflación, lea el apartado“Medición con la presión ajustada” en lapágina siguiente.

Al término de la medición, se visualizan laslecturas de las tensiones sistólica y diastólica yfrecuencia del pulso.
El brazalete expulsa el aire remanente y sedesinfla completamente.
Presione el botón START para apagar el aparato.
Después de un minuto de inactividad, el
aparato se apaga automáticamente.

Medición
progresiva

Presión sistólica
Presión diastólica
Clasificación según la OMS

Frecuencia del pulso

Nota: Deje que transcurran por lo menos tres minutos
entre mediciones de la misma persona.

Español

14

Evacuación automática
del aire remanente.

Mediciones
El modelo UA-1020-W está diseñado para detectar el pulso, inflando automáticamente
el brazalete hasta el nivel de presión sistólica.
Si la reinflación se produce de forma repetida, utilice los siguientes métodos.

Medición con la presión ajustada
Durante la medición de la presión sanguínea, puede producirse una reinflación.
Esto puede evitarse ajustando un valor de presión fijo.
1.

2.

En el paso 6 del procedimiento de ajuste delreloj (véase página 10), presione el botón
AJUSTAR para ir al modo de ajuste de presión.
El ajuste actual parpadea.
opara seleccionar dePresione el botón
entre los siguientes un valor de presión queesté unos 30 mmHg o más por encima de su
presión sistólica prevista.
AUTO : Presurización automática (valor predeterminado)
180 : Valor de presión de 180 mmHg (fijo)
210 : Valor de presión de 210 mmHg (fijo)
240 : Valor de presión de 240 mmHg (fijo)

3.

oApagarModo de ajuste
TriCheck.

Presione el botón AJUSTAR para ir al modo de ajuste TriCheck.
Presione el botón START para apagar el aparato. Después de tres minutos deinactividad, el aparato se apaga automáticamente.
La siguiente medición se realizará con el nuevo valor de presión.

Medición con la presión sistólica deseada
Utilice este método cuando la represurización se produzca
de forma repetida incluso aunque el valor de presión esté
ajustado en 240 en el procedimiento anterior, o bien cuando
los resultados no se muestren incluso aunque la presión
Al nivel deldisminuya a 20 mmHg o menos.
corazón

1.
2.

3.

Coloque el brazalete en el brazo (brazo
izquierdo preferiblemente).
Mantenga presionado el botón START
hasta que aparezca un número unos 30 a
40 mmHg más alto que su presión sistólica.
Suelte el botón START al alcanzar el valor
deseado, ya que la medición comenzará.
Luego continúe la medición de su presión
sanguínea como se describe en la página
anterior.
Español

15

Mantenga presionado STARTel botón
Suelte el botón cuando
se visualice la presión
sistólica deseada.

Para la medición, refiérase
a la página anterior.

Notas para mediciones correctas
Siéntese cómodamente. Coloque el brazo sobre una mesa con la palma hacia
arriba y el brazalete al mismo nivel que el corazón.
Relájese por unos cinco o diez minutos antes de iniciar la medición. Si seencuentra agitado o deprimido debido a un estrés emocional, la medición
reflejará este estrés con una lectura mayor (o menor) que la de su tensión arterial
normal y la lectura del pulso generalmente será más rápida que la normal.
La tensión arterial de una persona varía constantemente, dependiendo de lo queestá haciendo y lo que ha comido. Lo que tome podrá tener un efecto muy fuertey rápido sobre su tensión arterial.
Las mediciones de este aparato se basan en el latido del corazón. Si usted tiene
un latido muy débil o irregular, este aparato puede tener dificultad en determinarsu tensión arterial.
En caso de que el aparato detecte una condición anormal, el mismo detendrá
la medición e indicará un símbolo de error en la pantalla. Vea la página 6 parala descripción de los símbolos.
Este aparato está diseñado para su uso en adultos. Consulte con su médico
antes de usar este aparato en un niño. Un niño no debe usar este aparato sin la
atención de un adulto.
El rendimiento del tensiómetro automático puede verse afectado por temperatura
y humedad excesivas o por la altitud.

Medición TriCheckTM
El modo TriCheckTM realiza tres mediciones consecutivas de forma automática
y muestra los valores promedio de las tres mediciones.

Selección del modo TriCheckTM

En el modo de ajuste de presión,

1.

En el modo de ajuste de presión,presione el botón AJUSTAR para ir almodo de ajuste TriCheckTM. El ajuste
actual parpadea.

2.

Presione el botónopara ajustar
TMel modo TriCheck en ON (activado) oen OFF (desactivado).
ON: Modo TriCheckTM
OFF: Modo de medición normal (valor predeterminado)

3.

Presioneo
Cambiar el ajuste

Presione el botón START o AJUSTAR
para apagar el aparato. Después de tres
minutos de inactividad, el aparato se
apaga automáticamente.
Español

El ajuste actual parpadea.

o

16

START

Medición utilizando el modo TriCheckTM
1.

2.

Presione el botón START.
Aparecenbrevemente
todas
las
secciones de la pantalla. A continuación,
parece cero (0) parpadeando brevemente.
Luego comienza la primera medición.
Al término de la medición, se visualizan
las lecturas de las tensiones sistólica ydiastólica y frecuencia del pulso. A
continuación comienza una cuenta atrásde un minuto para la segunda medición.

3.

Transcurrido un minuto, comienza la
segunda medición.

4.

Al término de la medición, se visualizan
las lecturas promedio de las dos primeras
mediciones. A continuación comienza
una cuenta atrás de un minuto para la
tercera medición.

A nivel del corazón
START

Presione
Aparecen brevemente
todas las secciones
de la pantalla
Primera medición.

Un minuto después

5.

Transcurrido un minuto, comienza latercera medición.

6.

Al término de la medición, se visualizan laslecturas promedio de las tres mediciones yse almacenan en la memoria.

Segunda medición

Nota:

Durante la medición, aparece elsímbolo de TriCheckTM
Para cancelar la medición, pesione elbotón START. En este caso, no sealmacena ningún dato en la memoria.
Si durante la primera medición apareceel símbolo de error de ajuste delbrazalete, cancele la medición, vuelva
a colocar correctamente el brazalete e
inicie una nueva medición.
A Tras la medición, se almacenan en lamemoria los datos promedio de las tres
mediciones.
Si las mediciones se cancelan antes deque se complete la tercera medición,no se almacena ningún dato.
Español

Resultados de laprimera medición

Un minuto después

Promedio de las dos
primeras mediciones

Tercera medición

Promedio de lastres mediciones

17

Revisión de los datos de la memoria
Nota: Este aparato almacena en la memoria las últimas 90 mediciones.
Presioneo .
1. Presione el botón
Aparece el promedio de datos de todas
las mediciones y el número de datos.
(Si no hay datos, aparece “0”. Presione el botón,o START para apagar el aparato.)oPresión sistólica media
Presión diastólica media
Frecuencia media del pulso

Promedio detodos los datos

2. A continuación, cada vez que se presione elbotón (o el botón
para visualizar losdatos en el orden inverso), se muestran losdatos de la memoria de la siguiente forma.

Presión sistólica media
Presión diastólica media

Promedio de todas las mediciones AM (mañana)realizadas entre las 4:00 y las 9:59 horas.
(En el ejemplo, 10 mediciones. Si no hay
datos, aparece “--”.)

Frecuencia media del pulso
Promediode datos AM

Promedio de todas las mediciones PM(tarde/noche) realizadas entre las 18:00 y las
1:59 horas.
(En el ejemplo, 9 mediciones. Si no hay
datos, aparece “--”.)

Presión sistólica media
Presión diastólica media
Frecuencia media del pulso

Datos más recientes (No. n, en el
ejemplo, No. 35)
Los datos de la medición se muestrantres segundos después de visualizarse
el número de los datos.

Promediode datos PM

Sistólica
Diastólica

Último dato (No.1)
Los datos de la medición se muestrantres segundos después de visualizarse
el número de los datos.

Datos más recientes

3. Una vez que se haya mostrado el último dato,
presione el botón para volver a lavisualización del promedio de todas las
mediciones.
4. Presione el botón START para apagar elaparato. Después de un minuto de inactividad,
el aparato se apaga automáticamente.
Español

Frecuenciadel pulso

Sistólica
Diastólica
Último dato (más antiguo)

18

Frecuenciadel pulso¿ Qué es el indicador de LI/FibA?
Si el monitor detecta un ritmo irregular durante las mediciones, aparecerá elindicador LI/FibA en la pantalla con los valores de medición.

Nota: Le recomendamos que consulte a su médico si ve el símbolo
LI/FibA con frecuencia.

¿ A qué hace referencia FibA?
El corazón se contrae debido a corrientes eléctricas que surgen de él y envía
sangre por todo el cuerpo. La fibrilación auricular (FibA) se produce cuando las
corrientes eléctricas del atrio se confunden y ocasionan alteraciones en losintervalos del pulso. La FibA puede ocasionar estancamientos de sangre en elcorazón, lo que podría producir fácilmente coágulos de sangre, una de lascausas de las apoplejías y los infartos de corazón.

Indicador de barra de presión
El indicador monitoriza el progreso de la presión durante la medición.
Presurizando

Liberando aire

Inflación en progreso

Medición en progreso
Se ha completado la inflación

Indicador de clasificación de la OMS
Cada uno de los seis segmentos del indicadorde barra corresponde a la clasificación depresión sanguínea de la OMS que se describe
en la siguiente página.

Indicador de clasificación de la OMS
Hipertensión severa
Hipertensión moderada
Hipertensión leve
Normal alta
NormalOptimaEjemplo:

Hipertensión moderada

Hipertensión leve

: El indicador visualiza un segmento basado
en el dato actual, correspondiente a laclasificación de la OMS.

Español

19

Normal alta

Aspectos sobre la tensión arterial¿Qué es la tensión arterial?
La tensión arterial es la fuerza que ejerce la sangre contra las paredes de lasarterias. La tensión sistólica ocurre cuando el corazón se contrae, y la tensión
diastólica ocurre cuando el corazón se expande.
La presión sanguínea se mide en milímetros de mercurio (mmHg). La presión
arterial de una persona está representada por la tensión fundamental, que es laque se mide primero en la mañana cuando la persona está aún descansada yantes de tomar alimentos.

¿Qué es la hipertensión y cómo controlarla?
La hipertensión es la tensión arterial anormalmente alta, si no se corrige, puedecausar muchos problemas de salud, incluyendo ataques al corazón y apoplejía.
La hipertensión puede controlarse cambiando el estilo de vida, evitando el estrés ycon medicamentos bajo la supervisión de un médico.
Para evitar la hipertensión o para mantenerla bajo control:

No fume.
Reduzca la ingestión de sal y grasa.
Mantenga un peso apropiado.

Haga ejercicios con regularidad.
Sométase a exámenes físicos con
regularidad.

¿Por qué debe medirse la tensión arterial en la casa?
La medición de la tensión arterial en una clínica o un consultorio médico puede causar
temor y, por lo tanto producir una lectura elevada, de 25 a 30 mmHg mayor que lamedida en la casa. La medición en casa reduce los efectos de influencias externas
sobre las lecturas de tensión arterial, complementa las lecturas del médico yproporciona un historial más completo y exacto de la tensión arterial.

Clasificación de tensión arterial según la OMS
OMS ha establecido las normas parala evaluación de la tensión arterial, talcomo se muestra en el gráfico.

Variaciones de tensión
arterial
La tensión arterial de una persona
varía considerablemente en basediaria y estacional. La misma puedevariar en 30 a 50 mmHg dependiendode diversas condiciones durante eldía. En personas hipertensas, estas
variaciones son aun más notables.

Español20

La tensión arterial normalmente se eleva
cuando se está trabajando o jugando y baja asus niveles más bajos al dormir. Por lo tanto, nose preocupe demasiado de los resultados de
una sola medición.
Mídase la tensión arterial a la misma hora
todos los días usando el procedimiento
descrito en este manual y tenga conocimientode su tensión normal. Lecturas múltiples
proporcionarán un historial de tensión arterial
más completo. Asegúrese de anotar la fecha yhora al registrar su tensión arterial. Consulte con su médico para interpretar los datos
de su tensión arterial.

Localización y corrección de fallos
Problema
No aparece nada
en la pantalla, aun
cuando se conecta
la alimentación.

El brazalete no se
infla.

Causa probable
Las pilas se han agotado.
Los terminales de las pilas no
están en posición correcta.
El voltaje de las pilas está bajo.
Parpadea la marca(marcaLOW BATTERY). Si las pilas
están agotadas completamente,
no aparece esta marca.
El brazalete no está ajustado
debidamente.

Acción recomendada
Reemplace todas las pilas por
otras nuevas.
Coloque las pilas con sus
terminales negativo y positivo
coincidiendo con los indicadosen la banda inflable.
Reemplace todas las pilas por
otra nuevas.

Ajuste el brazaletecorrectamente.
Asegúrese de mantenerse muy
Movió su brazo o cuerpo durante
quieto y en silencio durante lala medición.
medición.
El aparato no mide
Siéntesecómodamenteyla presión arterial.
manténgase quieto. Coloque el
Las lecturas son
El brazalete no está posicionado
brazo sobre una mesa con lademasiado altas ocorrectamente.
palma hacia arriba y el brazaletedemasiado bajas.
al mismo nivel que el corazón.
Si el latido de su corazón es
muy débil o irregular, el aparato
tendrá dificultad en determinarsu tensión arterial.
El valor se diferencia al medido
Consulte el capítulo 3, ¿Poren una clínica o un consultorio
qué debe medirse la tensión
médico.
arterial en la casa?
Otros
Retire las pilas. Colóquelas
nuevamente de forma correcta e
intente la medición nuevamente.
Nota: Si las acciones descritas anteriormente no solucionan el problema, póngase en
contacto con su distribuidor. No abra el aparato ni intente repararlo, ya que
cualquier intento de reparación invalidará la garantía.

Español21

Mantenimiento
No abra el aparato, ya que éste contiene componentes eléctricos delicados y
una unidad de aire compleja que podrían dañarse. Si no puede solucionar el
problema aplicando las instrucciones de localización y corrección de averías,
solicite el servicio de reparación a su proveedor o al grupo de servicio de A&D.
El grupo de servicio de A&D proveerá información técnica, repuestos yunidades a los proveedores autorizados.
Este aparato está diseñado y fabricado para una vida de servicio prolongada.
No obstante se recomienda someter el monitor a la inspección cada 2 años afin de asegurar el funcionamiento y la precisión adecuadas. Sírvase ponerse
en contacto con su distribuidor autorizado o A&D para su mantenimiento.

Datos técnicos
Tipo:

UA-1020-W

Método de medición:

Medición oscilométrica

Rango de medición:

Tensión:0 a 299 mmHg

Tensión sistólica: 60 a 279 mmHg
Tensión diastólica: 40 a 200 mmHg
Precisión de medición:
Alimentación:

Pulso:

40 a 180 latidos/minuto

Tensión:±3 mmHg

Pulso:±5 %4 pilas de 1,5 V (R6P, LR6 o AA) o
un adaptador de CA (TB-233C) (No incluido)

Número de mediciones:

Aprox. 1000 veces
LR6 (pilas alcalinas)
Aprox. 300 veces
R6P (pilas de manganeso)
Con valor de presión de 180 mmHg a temperatura
ambiente de 23 °C

Clasificación:

Equipo ME alimentado internamente (suministrado por pilas)
Class II (suministrado por adaptador)
Modo de operación continuo

Prueba clínica:

De acuerdo con ISO81060-2: 2013
En el estudio de validación clínica, se determinó la presión
arterial diastólica en 85 sujetos mediante K5.

EMD:

IEC 60601-1-2: 2014

Almacenamiento de datos:

Las últimas 90 medidas

Condiciones ambientales
de funcionamiento
Condiciones de transporte /Almacenamiento

Temperatura de +10 a +40 °C / RH de 15 a 85 % /de 800 a 1060 hPa
Temperatura de -20 a +60 °C / RH de 10 a 95 % /Presión de 700 a 1060 hPa
Español22

Dimensiones:

Aprox. 140 [An] x 60 [Al] x 105 [Pr] mm

Peso:

Aprox. 285 g, excluyendozlas pilas

Protección de admisión

Aparato: IP20

Pieza aplicada

Brazalete

Vida útil

Aparato: 5 años (cuando se usa seis veces al día)
Brazalete: 2 años (cuando se usa seis veces al día)
Adaptador de CA: 5 años (cuando se usa seis veces al día)

Accesorio: Adaptador deCA, modelo TB-233C

El adaptador se conecta a la fuente de alimentación casera.
Sírvase tener contacto con la agencia A&D local para
comprarlo.
Es necesario inspeccionar o sustituir el adaptador CA
periódicamente.

Tipo BF

Símbolos impresos en el adaptador de CA
SímbolosFunción / Significado
Solo para uso en interiores
Aparato Clase II
Fusible térmico
FusibleEtiqueta de la directiva CE
Etiqueta de homologación EAC
Polaridad del enchufe del adaptador de CA
Accesorios vendidos por separado
BrazaleteNúmero de catálogo
Tamaño del brazaleteTamaño del brazo
CUF-F-LA
Brazalete grande de adulto
31 cm a 45 cm
CUF-I
Brazalete de amplio rango
22 cm a 42 cm
CUF-F-A
Brazalete de adulto
22 cm a 32 cm
CUF-F-SA
Brazalete pequeño de adulto
16 cm a 24 cm: Adaptador de CA
Número de catálogo
TB-233C

Clavija
Tipo C

Nota: Las especificaciones están sujetas a cambio sin previo aviso.
La clasificación IP hace referencia a los grados de protección otorgados por losmateriales contenedores según la norma CEI 60529. Este dispositivo está protegido
contra objetos extraños sólidos de 12 mm de diámetro y más, por ejemplo, un dedo.
Este dispositivo no está protegido contra el agua.

Español23

MEMO

Español

24

Indice
Introduzione .......................................................................... 2
Norme precauzionali.............................................................. 2
Note per il corretto utilizzo .................................................... 2
Identificazione delle parti ...................................................... 4
Simboli .................................................................................. 5
Modalità operative ................................................................. 7
Preparazione del misuratore .................................................. 8
Installazione / Sostituzione delle batterie ................................................... 8
Collegamento del bracciale allo strumento ................................................. 9
Collegamento dell’alimentatore a rete elettrica........................................... 9
Regolazione dell'orologio incorporato.......................................................10
Come scegliere il bracciale corretto ..........................................................11
Applicazione del bracciale .........................................................................11
Consigli per l’esecuzione di misurazioni corrette .....................................13
Durante la misura ......................................................................................13
Dopo la misura ..........................................................................................13

Misurazione della pressione ................................................. 14
Misurazione standard ................................................................................14
Misurazione con impostazione della pressione desiderata .......................15
Misurazione con impostazione della pressione di gonfiaggio desiderata .......15
Note per una misurazione corretta............................................................16

Misurazione TriCheckTM ....................................................... 16
Richiamo delle misure memorizzate .................................... 18
Cosa è l’indicatore IHB/AFib? ............................................... 19
Cosa è l’AFib? ...................................................................... 19
Indicatore dinamico della pressione ..................................... 19
Indicatore di classificazione dell’O.M.S................................. 19
Parlando di Pressione Arteriosa ............................................ 20
Cosa si intende per Pressione Arteriosa? ...................................................20
Cosa si intende per Ipertensione Arteriosa? ..............................................20
E come si controlla? ...................................................................................20
Perché è importante l’automisurazione al proprio domicilio? ...................20
Classificazione della pressione secondo le linee guida “O.M.S.” .........21

Localizzazione guasti .......................................................... 21
Manutenzione ...................................................................... 22
Dati tecnici .......................................................................... 22

Italiano1Introduzione
Gentile Cliente, la ringraziamo per aver acquistato il Misuratore Elettronicodella Pressione Arteriosa A&D. Concepito per una facilità di uso ed alta
precisione, il nostro prodotto la aiuterà a controllare la pressione arteriosa
quotidianamente.
Raccomandiamo di leggere con cura questo manuale prima di utilizzare il
misuratore per la prima volta.

Norme precauzionali
Il dispositivo è conforme alla Direttiva Europea CEE 93/42 per i Prodotti adutilizzo medico. Quanto sopra è evidenziato dal marchio di conformità.
(0123: Numero di riferimento dell'ente certificatore)
Il presente dispositivo deve essere utilizzato da parte di persone adulte, non dineonati o bambini.
Ambiente di utilizzo. Il dispositivo è idoneo per l’utilizzo da parte dell’utente inambienti di assistenza sanitaria domestici.
Il dispositivo è progettato per misurare la pressione sanguigna e la frequenza
cardiaca di soggetti umani a fini diagnostici.

Note per il corretto utilizzo
Il presente dispositivo è stato realizzato utilizzando componenti di precisione.
Si raccomanda di evitare condizioni estreme di temperatura, umidità, luce
solare diretta, urti o polvere.
Pulire il dispositivo e il bracciale utilizzando un panno morbido e asciutto o un
panno inumidito con acqua e detergente neutro. Non utilizzare mai alcool, benzene,
solventi o altri prodotti chimici aggressivi per pulire il dispositivo o il bracciale.
Evitare di piegare eccessivamente il bracciale o lasciare il tubo prolunga,
avvolto per troppo poiché si potrebbero danneggiare e comprometterel’attendibilità delle rilevazioni.
Fare attenzione a evitare lo strangolamento accidentale di bambini o infanti con il tubo
e il cavo.
Durante la misurazione non attorcigliare il tubo prolunga, in quanto potrebbero
verificarsi delle lesioni dovute alla pressione continua nel bracciale.
Il dispositivo ed il bracciale non sono impermeabili. Evitare il contatto diretto
con acqua o liquidi che potrebbero comprometterne la durata.
Il est possible que les mesures soient altérées si l’appareil est utilisé à proximité detéléviseurs, de fours à micro-ondes, de téléphones cellulaires, de sources derayonnement X ou d’autres appareils émettant de forts champs électriques.
I dispositivi per la comunicazione wireless, come i dispositivi delle reti domestiche, itelefoni cellulari, i telefoni cordless e le loro stazioni base, i walkie-talkie possono
interferire con il funzionamento di questo Monitor della Pressione Sanguigna. Èquindi opportuno tenerlo a una distanza minima di 30 cm da tali dispositivi.
Lo strumento, tutte le sue parti e le batterie esaurite non devono essere trattatecome normali rifiuti domestici, ma vanno smaltiti in conformità alle normative
ambientali in vigore.
Italiano 2

Quando si utilizza l'alimentatore a rete elettrica, assicurarsi che questo sia
facilmente scollegabile dalla presa elettrica in caso di necessità.
Quando si riutilizza il dispositivo, verificare che sia pulito.
Non modificare il dispositivo. L'apporto di modifiche può causare incidenti odanni al dispositivo stesso.
Pero misurare la pressione sanguigna, il braccio deve essere stretto dalbracciale a una pressione sufficiente a interrompere temporaneamente il flusso
sanguigno nell'arteria. Ciò può causare dolore, intorpidimento o lasciare
temporaneamente un segno rosso sul braccio, in particolare se la misurazione
viene ripetuta più volte di seguito.
Il dolore, l'intorpidimento o i segni rossi che possono eventualmente verificarsi
scompaiono dopo breve tempo.
Misurazioni della pressione sanguigna troppo frequenti possono provocare
danni al paziente a causa delle interferenze con il flusso sanguigno. In caso di
uso ripetuto del dispositivo, assicurarsi che il funzionamento del misuratore noncomporti una prolungata compromissione della circolazione sanguigna.
Se si è stati sottoposti a mastectomia, rivolgersi a un medico prima di utilizzare
il dispositivo.
Evitare che i bambini usino il dispositivo senza sorveglianza e tenere il dispositivo
fuori dalla loro portata. In caso contrario, si potrebbero causare incidente o danni.
Presenta piccole parti che possono causare soffocamento se ingerite
inavvertitamente dai bambini.
Scollegare l’adattatore C.a. quando non lo si usa durante la misurazione.
L’utilizzo di accessori non indicati in questo manuale potrebbe compromettere
la sicurezza.
In caso di cortocircuito della batteria, la suddette potrebbe riscaldarsi e causare
potenzialmente delle ustioni.
Consentire al dispositivo di adattarsi all’ambiente circostante, prima dell’uso(circa un’ora).
Non sono stati condotti test clinici sui neonati e le donne in gravidanza. Nonusare su neonati o donne in gravidanza.
Non toccare contemporaneamente le batterie, la presa jack C.c. e il paziente.
Si potrebbero causare scosse elettriche.
Non gonfiare senza avvolgere il bracciale attorno al braccio.

Controindicazioni
Di seguito sono indicate le precauzioni da osservare per il corretto utilizzo deldispositivo.
Non applicare il bracciale a un paziente già collegato a un altro dispositivo
medico. Il dispositivo potrebbe non funzionare correttamente.
Le persone affette da grave deficit circolatorio al braccio devono consultare un
medico prima di utilizzare il dispositivo, al fine di evitare problemi medici.
Non eseguire auto-diagnosi dei risultati delle misurazioni, né iniziare trattamenti
in autonomia. Per la valutazione dei risultati ed eventuali trattamenti, rivolgersi
sempre al medico.
Non applicare il bracciale se il braccio presenta ferite non cicatrizzate.
Non applicare il bracciale in caso di terapia con flebo o trasfusioni di sangue
eseguite al braccio del paziente. Pericolo di lesioni o incidenti.
Non utilizzare il dispositivo in ambienti in cui siano presenti gas infiammabili,
quali i gas anestetici. Pericolo di esplosione.
Non utilizzare il dispositivo in ambienti con concentrazioni molto elevate diossigeno, quali camere iperbariche o tende a ossigeno. Pericolo di incendio oesplosione.
Italiano 3

Identificazione delle parti
Jack a CC
Display a cristalli liquidi

Pulsante
Pulsante
Pulsante
CONFIGURAZIONE

Spina innesto aria

Pulsante START(AVVIO)

Presa d’aria
Tubo prolunga
Indicatore di scelta delbracciale adeguato

Simbolo di indicazione
posizione arteria brachiale
Bracciale

Indicatore di posizione
Coperchio vano batterie
Vano batterie

Batterie da 1,5 V(R6P, LR6 o AA)

Particolaridel Display

MEMORIA

Valori medi

Impostazioni di pressione

Simbolo IHB/AFib
Simbolo di errore diposizionamento del bracciale
Pressione sistolica

Simbolo di errore di movimento
Simbolo TriCheckTM
Simbolo del cuore(Indicazione misura in corso)

Pressione diastolica

Indicatore di classificazione OMS e
indicatore di pressione

Frequenza cardiaca

Indicatore batterie
Simbolo AM / PM
Italiano 4

Orologio

Simboli
Simboli impressi sulla confezione del dispositivo e sull'alimentatore a rete elettrica
Simboli

SN

EC

REP

IP

Funzione/Significato
Posizionare l’apparecchio in modalità
standby, quindi collegarlo
Guida all'installazione delle batterie
Corrente continua
Numero di matricola
Data di produzione
Tipo BF: dispositivo, bracciale e tubisono stati progettati al fine di garantire
una protezione particolare contro
eventuali scosse elettriche

Azione raccomandata

Etichetta Direttiva CE sui dispositivi medici
Etichetta Direttiva RAEE (direttiva
Europea di smaltimento dei dispositivi
elettrici ed elettronici)
Produttore
Rappresentante UE
Consultare il manuale/l'opuscolo di istruzioni
Polarità del jack a CC
Simbolo internazionale di protezione
Tenere ascitto

Simboli visualizzati sul display
SimboliFunzione/Significato
Indicazione visualizzata durante la fasedi misurazione. Lampeggia durante il
rilevamento della frequenza cardiaca.

Azione raccomandata
La misurazione è in corso.
Rimanere quanto più
possibile fermi e rilassati.

Simbolo IHB/AFib
Indicazione visualizzata se viene rilevato un
battito cardiaco irregolare. Potrebbe
illuminarsi se viene rilevata una leggerissima
vibrazione come un brivido o un tremore.

La modalità TriCheckTM consente dieseguire automaticamente tre
misurazioni consecutive e divisualizzarne i valori medi.

Simbolo visualizzato se viene
rilevato un movimento corporeo odel braccio.
Italiano 5Laletturapotrebbe
presentare un valore
incorretto.
Ripetere la misurazione.
Rimanere quanto possibile
fermi e rilassati.

Simboli

Funzione/Significato
Simbolo visualizzato in fase dimisurazione se il bracciale è allentato.

Azione raccomandata
La lettura potrebbe
presentare un valore
incorretto.
Applicare e stringere
adeguatamente il bracciale
e ripetere la misurazione.

Misurazioni precedenti
MEMORIZZATE.
Valori medi
BATTERIE CARICHE
Indicatore di stato di carica dellebatterie durante la misurazione.
Qualora l’indicatore
lampeggi, sostituire tutte lebatterie.
Ripetere la misurazione.
Valori della pressione non corretti a causa di
Rimanere quanto possibile
movimenti durante la misurazione.
fermi e rilassati.
Differenza tra pressione sistolica ediastolica inferiore a 10 mmHg.
BATTERIE SCARICHE
Indicatore di batterie in esaurimento.

Pressione di gonfiaggio non sufficiente.
Il bracciale non è applicato in modo corretto.
ERRORE DI VISUALIZZAZIONE DEL
BATTITO CARDIACO

Applicare e stringere
adeguatamente il
bracciale e ripetere la
misurazione.

Il battito cardiaco non è stato rilevato inmodo corretto.
Errore interno del misuratore dipressione arteriosa
SYS
DIA
PUL
AM
PM

Pressione arteriosa sistolica in mmHg
Pressione arteriosa diastolica in mmHg
Battiti cardiaci al minuto
Dati rilevati tra le 4:00 e le 9:59.
Dati rilevati tra le 18:00 e le 1:59.
Impostazioni di pressione
Indica il valore della pressione
precedentemente impostato dall’utente.

Italiano 6

Rimuovere le batterie epremere il pulsante
START (AVVIO), quindi
reinstallare le batterie. Se
l’errore persiste, contattare
il rivenditore.

Modalità operative
1. Misurazione normale con memorizzazione dei dati
Premere il pulsante START. La pressione arteriosa viene misurata e i dati sono
memorizzati nel dispositivo. Sono memorizzati automaticamente gli ultimi 90
valori di pressione.

2. Richiamo dei dati

Premere il pulsanteo
per richiamare i
dati.
Vengono visualizzati i valori medi di tutte lemisurazioni, come mostrato nella figura adestra.

Quindi, ogni volta che si preme il pulsante, le misure memorizzate vengono
visualizzate come segue.
Media di tutte le misurazioni AM (antimeridiane)
rilevate tra le ore 4:00 e 9:59.
Media di tutte le misurazioni PM (postmeridiane)
rilevate tra le ore 18:00 e 1:59.
Dati più recenti (No.n, nell’esempio, No.35)
Ultimi dati (No.1)
Per i dettagli su come richiamare i dati,
consultare “Richiamo delle misurememorizzate”. ( Consultare pagina 18.)

3. Per eliminare tutti i dati memorizzati
Premere entrambi i pulsantie . Vengono
visualizzati il simbolo
e l’indicatore batterie. Per
eliminare tutti i dati memorizzati, premere e tenere
premuti entrambi i pulsantiefinché il simbolo
luminoso
inizia a lampeggiare.

4. Misurazione con impostazione della pressione di gonfiaggio
desiderata
Vedere descrizione completa a pagina 15.
Italiano 7

Preparazione del misuratore
Installazione / Sostituzione delle batterie
1. Rimuovere il coperchio del vano batterie.
2. Rimuovere le batterie esaurite ed inserire
le nuove batterie, rispettando le polarità (+e -) come indicato nella figura a lato.
Utilizzare esclusivamente batterie R6P,
LR6 o AA.
3. Chiudere il coperchio del vano batterie.

Procedura 1
Procedura 3

Procedura 1

Procedura 2

ATTENZIONE
Inserire le batterie come indicato nel vano porta
batterie. In caso contrario, l'apparecchio non
funzionerà.
Quando il simbolo(BATTERIE SCARICHE) lampeggia sul display, sostituire
tutte le batterie. Non mischiare batterie nuove con batterie usate, per evitare diridurre la durata delle batterie o causare malfunzionamenti del dispositivo.
Spento il dispositivo, attendere almeno due secondi per sostituire le batterie.
(BATTERIE SCARICHE) viene visualizzato anche dopo averSe il simbolo
sostituito le batterie, eseguire una misurazione della pressione arteriosa. Il
dispositivo è in seguito in grado di riconoscere le batterie nuove.
Il simbolo(BATTERIE SCARICHE) non viene visualizzato in caso di batteriecompletamente esaurite.
La durata delle batterie varia in base alla temperatura ambiente e potrebbe
risultare ridotta in caso di esposizione a basse temperature. In genere, quattro
batterie LR6 nuove durano circa un anno, quando vengono utilizzate due volteogni giorno per la misurazione
Utilizzare solamente le batterie del tipo specificato. Le batterie fornite indotazione al dispositivo sono a solo scopo di prova delle prestazioni del monitor ela loro durata potrebbe essere breve.
Togliere le batterie nel caso in cui il dispositivo non sia utilizzato per lungo tempo.
Le batterie potrebbero infatti essere soggette a perdite, causando quindi dei
malfunzionamenti.

Italiano 8

Preparazione del misuratore
Collegamento del bracciale
allo strumento

Presa d’aria

Inserire fermamente la spina innesto
aria nella presa d’aria.

Spina innesto aria

Collegamento dell’alimentatore a rete elettrica
Inserire la spina dell’alimentatore nel jack a CC.

Inserire successivamente l’alimentatore
nella presa elettrica.
Utilizzare l’alimentatore a rete elettrica
specificato.( Consultare pagina 23.)

Spina dell’alimentatore

Jack a CC

Per scollegare l’alimentatore dallapresa elettrica, afferrare e rimuovere il
corpo dell’alimentatore dalla presa.
Per scollegare la spina dell’alimentatore dal
Misuratore Elettronico Della Pressione Arteriosa,afferrare ed estrarre la spina dell’alimentatore
dal misuratore.

Italiano 9

Preparazione del misuratore
Regolazione dell'orologio incorporato

PremereRegolare l'orologio prima dell'uso.
1. Premere il pulsante CONFIGURAZIONE finché
le cifre iniziano a lampeggiare.

Anno

o

o , selezionare l'anno.
2. Agendo sul pulsantePremere il pulsante CONFIGURAZIONE per
impostare l'anno attuale e conseguentemente
per selezionare il mese ed il giorno. La data puòessere impostata tra gli anni 2010 e 2059

Meseo , selezionare il mese.
3. Agendo sul pulsantePremere il pulsante CONFIGURAZIONE per
impostare il mese attuale e conseguentemente
per selezionare il giorno.

Giornoo , selezionare il giorno.
4. Agendo sul pulsantePremere il pulsante CONFIGURAZIONE per
impostare il giorno attuale e conseguentemente
per selezionare l'ora ed i minuti.

o

o

Ora

o , selezionare l'ora.
5. Agendo sul pulsantePremere il pulsante CONFIGURAZIONE per
impostare l'ora attuale e conseguentemente perselezionare i minuti.
o , selezionare i minuti.
6. Agendo sul pulsante(Premere il pulsante CONFIGURAZIONE perpassare alla modalità di impostazione dellapressione. Per i dettagli, consultare pagina 15).
Premere il pulsante START per spegnere il
dispositivo.
Nota:

Dopo tre minuti di inattività, il dispositivo
si spegnerà automaticamente.
Quando non è stato impostato,
sull’orologio figurano le lineettemostrate sulla destra.
Tenendo premuto il pulsanteosiottiene una successione consecutiva dei
valori.

Italiano

10ooMinutiSpegnimento

Premendo il pulsante
START il dispositivo si
spegne in qualunquemomento.

Modalità di impostazionedella pressionePreparazione del misuratore
Come scegliere il bracciale corretto
L’uso della dimensione corretta di un bracciale è importante per una lettura
precisa. Se il bracciale non presenta la dimensione corretta, la lettura potrebbe
produrre un valore incorretto della pressione sanguigna.
La dimensione del braccio è stampata su ogni bracciale.
Il simbolo
e l'indicatore di scelta del bracciale adeguato determinano se è
stato utilizzato il bracciale corretto. (Consultare "Simboli stampati sul bracciale "alla pagina seguente.)
Se il simbolo
è posizionato fuori dal range, si raccomanda l’acquisto di un
bracciale appropriato (consultare la tabella sottostante).
Il bracciale è un prodotto di normale consumo. In caso di usura, acquistarne uno nuovo.
Dimensione del braccio
31 cm a 45 cm
22 cm a 42 cm
22 cm a 32 cm
16 cm a 24 cm

Dimensione raccomandata delbracciale
Bracciale grande per adulti
Bracciale ampio
Bracciale per adulti
Bracciale piccolo per adulti

Numero di catalogo
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Misura del braccio: corrisponde alla circonferenza del bicipite.

Applicazione del bracciale

Arrotolare la manica evitando
di stringere troppe il braccio

1. Avvolgere il bracciale intorno al
braccio superiore, a circa 1 - 2 cm
sopra la piega del gomito, comeindicato nella figura a lato. Applicare il
bracciale direttamente a contatto conla pelle, in quanto i vestiti possono
causare errori nella misura.

Bracciale
1-2 cmTubo prolunga
Simbolo di indicazione
posizione arteria brachiale

2. La compressione al braccio causata
da un'avvolgimento troppo strettodella manica della camicia, puo`
causare una misura errata.
3. Verificare che il simbolo
all'interno del range di selezione.

Avvolgere il bracciale
saldamenteèSimboloe Indicatore di sceltadel bracciale adeguato

Italiano11

Preparazione del misuratore
Simboli stampati sul bracciale
Simboli●▲
REF

Funzione/Significato

Azione raccomandata

Simbolo di indicazione posizione arteria
brachiale

Posizionare il simbolo ●sull’arteria del braccio o inallineamento con il dito anulare
nella parte interna del braccio.

IndiceNumero di catalogo
Gamma adatta a bracciale per adulti.
È stampato sul bracciale per adulti.
Gamma superiore stampata sul bracciale Utilizzare il bracciale grande
per adulti anziché il bracciale
per adulti e sul bracciale ampio.
per adulti o il bracciale ampio.
Gamma adatta a bracciale ampio.
È stampato sul bracciale ampio.
Gamma adatta a bracciale grande per adulti
È stampato sul bracciale grande per adulti
Gamma inferiore stampata sul bracciale per Utilizzare il bracciale piccolo
per adulti anziché il bracciale
adulti e sul bracciale ampio.
per adulti o il bracciale ampio.
Gamma adatta a bracciale piccolo per
adulti.
È stampato sul bracciale piccolo per adulti.
Utilizzare il bracciale per adulti
È stampata sul bracciale grande per adulti anziché il bracciale grande per
e sul bracciale piccolo per adulti.
adulti o il bracciale piccolo per
adulti.
Numero di lotto
Indicatore di scelta del bracciale adeguato

Bracciale grande
per adulti

AL

Bracciale ampio

S

W

L

Bracciale per adulti

S

A

L

S

A

Bracciale piccolo
per adulti

Italiano12Preparazione del misuratore
Consigli per l’esecuzione di misurazioni correttePer garantire la migliore precisione delle rilevazioni suggeriamo di:
Sedersi comodamente su una sedia. Appoggiare il braccio sul tavolo. Non
incrociare le gambe. Tenere i piedi appoggiati al suolo e raddrizzare il dorso.
Prima di procedere alla misurazione rilassarsi per 5 -10 minuti.
Avvolgere il bracciale allo stesso livello del cuore.
Durante la misurazione, mantenersi fermi e tranquilli.
Non effettuare la misurazione subito dopo aver eseguito esercizi fisico o aver fatto ilbagno. In tali casi, riposare per venti-trenta minuti prima di procedere alla misurazione.
Si consiglia di misurare la pressione arteriosa tutti i giorni alla stessa ora

Durante la misura

Durante la misurazione, è normale sentire il bracciale a pressione stringere
fortemente (Non preoccuparsi).

Dopo la misura

Dopo la misurazione, premere il pulsante di START per spegnere il dispositivo.
Rimuovere il bracciale e registrare i vostri dati. Dopo un minuto di inattività, il
dispositivo si spegnerà automaticamente.

Italiano13

Misurazione della pressione
Prima di eseguire la misurazione, consultare “Note per una misurazione corretta” a pagina
16.

Misurazione standard
1.

Applicare il bracciale preferibilmente sul braccio sinistro
allo stesso livello del cuore. Durante la misurazione
rimanere quanto più possibile fermi e rilassati.

2.

Premere il pulsante START .
Per un attimo verranno visualizzati tutti i segmenti sul
display e in un secondo tempo, come mostrato nella
figura a lato, verrà visualizzato lo 0 (zero)lampeggiante per alcuni istanti.
Successivamente il bracciale inizierà a gonfiarsi (ènormale sentire stringere decisamente) e unindicatore dinamico a barre sarà visualizzato nell'area
a sinistra del display, mostrando l’incremento dellapressione di gonfiaggio.
Nota:

3.

Una volta completata la fase di pressurizzazione lo
sgonfiamento sarà automatico ed il simbolo(simbolo del cuore) inizierà a lampeggiare,
indicando che la misurazione è in corso. Una volta
rilevate le pulsazioni, il simbolo del cuore
lampeggerà in sincronia con il battito cardiaco.
Nota:

4.

5.

Nel caso in cui, in qualsiasi momento, si
desideri sospendere la misurazione, premere
nuovamente il pulsante START .

PremereSTART

Tutte le parti deldisplay vengono
visualizzate
Visualizzazione
zero
Inizio della fase digonfiamento

Pressurizzazione

Nel caso in cui non si ottenga una pressione
di gonfiaggio adeguata, il misuratore inizieràautomaticamente a rigonfiare il bracciale.
Per evitare il rigonfiaggio, consultare“Misurazione con impostazione dellapressione desiderata” alla pagina successiva.

Una volta completata la misurazione, saranno
visualizzati i valori relativi alla pressione sistolica,
diastolica e alla frequenza cardiaca, insieme a unsegmento lampeggiante in corrispondenza della scaladei valori dell’ O.M.S. Il bracciale espellerà l’aria
residua e si sgonfierà completamente.
Premere il pulsante START per spegnere il dispositivo.
Dopo un minuto di inattività, il dispositivo si spegnerà
automaticamente.

Nota:

Al livello delcuore

E’ consigliabile lasciar passare almeno 3 minuti prima
di effettuare una nuova misura sulla stessa persona.

Italiano

14

Misurazione
in corso

Pressione sistolica
Pressione diastolica
Classificazione OMS
Frequenza cardiaca
Espulsione automatica
dell’aria residua nel braccialeMisurazione della pressione
Il modello UA-1020-W è stato progettato per rilevare in modo completamente
automatico la pressione e la frequenza cardiaca.
Se il gonfiaggio si ripete, applicare i metodi indicati di seguito.

Misurazione con impostazione della pressione desiderata
Durante la fase di misurazione della pressione arteriosa potrebbe azionarsi il rigonfiaggio.
Per evitare il rigonfiaggio, è possibile impostare un valore fisso di pressione.
1.

2.

Al punto 6 della procedura di regolazione
dell’orologio (consultare pagina 10), premere ilpulsante CONFIGURAZIONE per passare alla
modalità di impostazione della pressione.
L’impostazione attuale lampeggia.

ooper selezionare un
Premere il pulsantevalore di pressione di 30 mmHg circa osuperiore a quello della pressione sistolica
prevista tra i valori che seguono.
AUTO : Pressurizzazione automatica (valore predefinito)
180 : Valore di pressione di 180 mmHg (fisso)
Spegnimento Modalità diimpostazione
210 : Valore di pressione di 210 mmHg (fisso)
TriCheck.
240 : Valore di pressione di 240 mmHg (fisso)3.

Premere il pulsante CONFIGURAZIONE per passare alla modalità di impostazione
TriCheck. Premere il pulsante START per spegnere il dispositivo. Dopo tre minuti diinattività, il dispositivo si spegnerà automaticamente.
La misurazione successiva verrà eseguita con il nuovo valore di pressione.

Misurazione con impostazione della pressione di gonfiaggio desiderata
Applicare questo metodo se il rigonfiaggio si ripete malgrado l’impostazione su 240 delvalore di pressione nella procedura di cui sopra, o se i risultati non vengono visualizzati
malgrado la riduzione della pressione a 20 mmHg o meno.
1.

2.

3.

Applicare il bracciale preferibilmente sul
braccio sinistro allo stesso livello del cuore.
Durante la misurazione rimanere quanto più
possibile fermi e rilassati.
Premere e tenere premuto il pulsante
START finché sarà visualizzato un
numero tra 30 e 40 mmHg superiore aquello della pressione sistolica prevista.
Dopo aver raggiunto il valore desiderato,rilasciare il pulsante .START, la misurazione
inizierà. Proseguire quindi con la
misurazione della pressione sanguigna
come descritto alla pagina precedente.
Italiano

15

A livello del cuore

Premere e tenere premuto il
pulsante START

START

Alla visualizzazionedella pressione digonfiaggio richiesta,
rilasciare il pulsantePer quanto riguarda la procedura di
rilevazione della pressione, fareriferimento alla pagina precedente

Note per una misurazione corretta
Sedersi in una posizione comoda. Appoggiare il braccio su un tavolo con ilpalmo della mano rivolto verso l’alto e con il bracciale alla stessa altezza delcuore.
Rilassarsi per circa 5 - 10 minuti prima di effettuare una misurazione. Se si èagitati o emozionati, la misurazione rifletterà questa situazione, con un valoredecisamente più alto (o più basso) della pressione arteriosa normale. Anche lafrequenza cardiaca sarà solitamente più alta del normale.
La pressione arteriosa di un soggetto varia costantemente, secondo l’attività
fisica ed emozionale, l’assunzione di cibi e la luce. Anche l’assunzione dibevande alcoliche può avere un effetto sull’andamento dei valori di pressione
arteriosa.
Questo apparecchio si basa sul principio di misura oscillometrico, quindi ingrado di valutare e analizzare i parametri delle pulsazioni cardiache. Se i battiti
sono molto deboli o irregolari, l’apparecchio potrebbe avere difficoltà adeterminare la pressione arteriosa.
Se l'apparecchio rileva una circostanza anormale, arresterà la misurazione e
visualizzerà un simbolo di errore. (Vedere pagina 6 per la descrizione dei
simboli).
Questo dispositivo deve essere gestito da persone adulte. Consultare il proprio
medico prima di utilizzare lo strumento su un bambino. In ogni caso, l’utilizzo su
bambini deve avvenire sempre sotto la stretta sorveglianza di persone adulte.
La prestazioni del monitor automatico per la pressione arteriosa potrebbero
essere influenzata dalla temperatura o umidità eccessiva o l'altitudine.

Misurazione TriCheckTM
La modalità TriCheckTM consente di eseguire automaticamente tre misurazioni
consecutive e di visualizzarne i valori medi.

Come scegliere la modalità TriCheckTM
1.

Nella modalità di impostazione dellapressione,
premereilpulsante
CONFIGURAZIONE per passare alla
modalità di impostazione TriCheckTM.
L’impostazione attuale lampeggia.

2.

oper ON
Premere il pulsante(attivare) o OFF (disattivare) la modalità
TriCheckTM.
ON:
Modalitàdiimpostazione
TM
TriCheck
OFF: Modalità di misurazione standard (valore predefinito)

3.

premerePremereilpulsante
STARTo
CONFIGURAZIONE per spegnere il
dispositivo. Dopo tre minuti di inattività, il
dispositivo si spegnerà automaticamente.
Italiano

Nella modalità di impostazionedella pressione,

L’impostazione attuale
lampeggia.
oModificare l’impostazione
o

16

START

Misurazione in modalità TriCheckTM
1.

2.

Premere il pulsante START.
Per un attimo verranno visualizzati tutti isegmenti sul display e in un secondo tempo,
come mostrato nella figura a lato, verrà
visualizzato lo 0 (zero) lampeggiante per alcuni
istanti. Successivamente viene avviata la prima
misurazione.
Una volta completata la misurazione, saranno
visualizzati i valori relativi alla pressione sistolica,
diastolica e alla frequenza cardiaca, insieme aun segmento lampeggiante in corrispondenzadella scala dei valori dell’ O.M.S. Quindi, inizia
un conto alla rovescia di un minuto che precede
la seconda misurazione.

3.

Trascorso un minuto, viene avviata laseconda misurazione.

4.

Una volta completata la misurazione, vengono
visualizzate le letture medie della prima edella seconda misurazione. Quindi, inizia unconto alla rovescia di un minuto che
precede la terza misurazione.

5.

Trascorso un minuto, viene avviata la terza
misurazione.

Al livello del
cuore
START

PremereTutte le parti del display
vengono visualizzate
Prima misurazione

Un minuto dopo

Risultati dellaprima misurazione

Seconda misurazione
6.

Una volta completata la misurazione, vengono
visualizzate e memorizzate le letture mediedelle tre misurazioni.

Nota:
Durante la fase di misurazione, vieneTMvisualizzato il simbolo TriCheck .
Per annullare la misurazione, premere ilpulsante START. In tal caso, i dati non
vengono memorizzati.
Se durante la prima misurazione viene
visualizzato il simbolo di errore diposizionamento del bracciale, annullare
la misurazione, applicare di nuovo ilbracciale correttamente e avviare una
nuova misurazione.
Dopolamisurazione,
vengonomemorizzati i valori medi delle tre
misurazioni.
Durante l’annullamento delle misurazioni,
non verrà memorizzato alcun dato prima
che la terza misurazione abbia fine.
Italiano

Un minuto dopo

Media della prima e dellaseconda misurazione

Terza misurazione

Medie delle tre
misurazioni

17

Richiamo delle misure memorizzate
Nota: Il dispositivo memorizza automaticamente gli ultimi 90 valori di pressione.
Premere1. Premere il pulsanteo .
Vengono visualizzati i valori medi ditutte le misurazioni e il numero dei dati.
(In assenza di dati, viene visualizzato “0”.
,o START per
Premere il pulsante
spegnere il dispositivo.)oPressione sistolica media
Pressione diastolica media
Frequenza cardiaca media

Media ditutti i dati

2. Quindi, ad ogni pressione del pulsante(o il pulsanteper visualizzare i dati inordine inverso), vengono visualizzati i dati
in memoria come segue.

Pressione sistolica media

Media di tutte le misurazioni AM(antimeridiane) rilevate tra le ore 4:00 e 9:59.
(Nell’esempio, 10 misurazioni, In assenza didati, viene visualizzato “--”.)

Pressione diastolica media
Frequenza cardiaca media
Media dei
dati AMMedia di tutte le misurazioni PM(postmeridiane) rilevate tra le ore 18:00 e 1:59.
(Nell’esempio, 9 misurazioni, In assenza didati, viene visualizzato “--”.)

Pressione sistolica media
Pressione diastolica media
Frequenza cardiaca media

Dati più recenti (No.n, nell’esempio, No.35)
Tre secondi dopo la visualizzazione delnumero dei dati, vengono visualizzati i datidella misurazione.

Media dei
dati PM

Sistolica

Ultimi dati (No.1)
Tre secondi dopo la visualizzazione delnumero dei dati, vengono visualizzati i datidella misurazione.

Dati più recenti

3. Dopo aver visualizzato gli ultimi dati,
per tornare alla
premere il pulsantevisualizzazione della media di tutte lemisurazioni.

Sistolica

4. Premere il pulsante START per spegnere
il dispositivo. Dopo un minuto di inattività,
il dispositivo si spegnerà
automaticamente.
Italiano

Diastolica
Frequenza
cardiaca

Ultimi dati (Dati più vecchi)

18

Diastolica
Frequenza
cardiaca

Cosa è l’indicatore IHB/AFib?
Quando il monitor rileva un ritmo irregolare durante le misurazioni, l’indicatore
IHB/AFib sarà visualizzato sul display unitamente ai valori di misurazione.
Nota: Raccomandiamo di contattare il proprio medico se viene indicato
frequentemente questo simbolo IHB/Afib.

Cosa è l’AFib?
Il cuore si contrae a causa di segnali elettrici che hanno luogo nel cuore e invia
sangue in tutto il corpo. La fibrillazione atriale (AFib) si verifica quando ilsegnale elettrico nell’atrio diventa confuso e causa disturbi nell’intervallo dellepulsazioni. L’AFib può causare ristagni del sangue nel cuore, i quali possono
causare facilmente coaguli di sangue, una delle cause di ictus e infarto.

Indicatore dinamico della pressione
Lo strumento è provvisto di un indicatore dinamico a barre che controlla il progressodella pressione durante la fase di misurazione. L’indicatore è visibile nella partesinistra del display e, si incrementa e diminuisce progressivamente durante il
gonfiaggio del bracciale e durante la misura della pressione.
Pressurizzazione
Depressurizzazione

Gonfiamento in corso

Misurazione in corso
Gonfiamento

Indicatore di classificazione dell’O.M.S.
L’Organizzazione Mondiale della Sanità (O.M.S.) hasancito delle linee guida per la corretta valutazione einterpretazione dei valori di pressione misurata.
Ha suddiviso in 6 stadi la classificazione dei valori,come evidenziato dalla tabella sottostante:

Indicatore di classificazione O.M.S.
Ipertensione severa (grado 3)
Ipertensione moderata (grado 2)
Ipertensione lieve (grado 1)
Pressione normale – alta
Pressione normale
Pressione ottimale

Esempio:

Ipertensione moderata

Ipertensione lieve: L’indicatore mostra un segmento sulla base dei dati
rilevati, corrispondente alla
Classifica zione O.M.S

Pressione normale – alta
Italiano19

Parlando di Pressione Arteriosa
Cosa si intende per Pressione Arteriosa?
La pressione arteriosa è la forza esercitata dal sangue contro le pareti delle arterie.
La pressione sistolica (massima) si verifica nel momento in cui il cuore si contrae espinge il sangue nelle arterie. La pressione diastolica (minima) si verifica quando il cuore
si dilata, ricevendo il sangue all’interno del ventricolo sinistro, e si prepara a una nuova
contrazione. La pressione arteriosa è misurata in millimetri di mercurio (mmHg).

Cosa si intende per Ipertensione Arteriosa?
Secondo le linee guida dell’Organizzazione Mondiale della Sanità, si può parlare diipertensione arteriosa quando si hanno valori di pressione sistolica e diastolica
persistentemente elevati oltre il limite di 140 / 90 mmHg. E’ importante però non fareriferimento esclusivamente ad una singola misura, ma è necessario considerare anche ilvalore di più rilevazioni in un periodo di tempo stabilito dal medico ed anche lefluttuazioni della pressione stessa. Recenti studi a lungo termine hanno dimostrato che ilimiti da considerare per la pressione domiciliare automisurata, sono infatti meno elevati
e si attestano su valori di 135 mmHg per la pressione sistolica e 85 mmHg per lapressione diastolica.

E come si controlla?
E’ possibile tenere sotto controllo l’ipertensione arteriosa variando il proprio stile di vita ealimentare, evitando il più possibile situazioni di stress, oppure con opportuni trattamenti
farmacologici secondo prescrizione medica. Per prevenire l'ipertensione arteriosa o pertenerla sotto controllo, le linee guida dell’ O.M.S. consigliano di:Non fumare
Eseguire esercizi a cadenza regolare
Ridurre l’uso di sale e moderare
Sottoporsi a check-up regolaril’assunzione di grassi animali
Mantenere il proprio peso a livelli
ottimali

Perché è importante l’automisurazione al proprio domicilio?
La pressione che il medico misura mediante lo sfigmomanometro tradizionale è quasi
sempre caratterizzata da un rialzo generato dall’emozione che colpisce il paziente al
momento della visita medica, rialzo la cui entità e durata sono variabili da soggetto asoggetto. Questo fattore è comunemente conosciuto come ”ipertensione da camicebianco” o più opportunamente come “ipertensione clinica isolata” e si manifesta a causa
degli stati di ansia generati dalla presenza del medico o della struttura sanitaria.
Un vantaggio della misurazione domiciliare è che in genere è esente da questo
fenomeno. Un altro vantaggio dell’automisurazione è che permette di ottenere più valori
nell’arco di diversi giorni o settimane, con una stima precisa delle variazioni dellapressione.
La misurazione domiciliare inoltre può fornire utili informazioni nella valutazione deltrattamento antiipertensivo. In questo caso la pressione domiciliare può rivelarsi un
metodo molto efficace, perché il suo impiego coinvolge attivamente il paziente con effetti
potenzialmente positivi sull’aderenza alla terapia.

Italiano20Classificazione della pressione secondo le linee guida “O.M.S.”
Gli standard per la valutazione della pressione
arteriosa, indipendentemente dall’età, sono stati
stabiliti dall’Organizzazione Mondiale della Sanità(O.M.S.), come mostrato nella tabella a fianco.
Normalmente, la pressione arteriosa aumenta
durante l’attività quotidiana e il lavoro e, tende adiminuire durante il sonno. Le linee guida inoltre
consigliano di effettuare quotidianamente lemisurazioni possibilmente sempre alla stessa ora.
L’esecuzione di molte rilevazioni permette diottenere un quadro della propria pressione arteriosa
più completo. E’ sempre meglio annotare la data el’ora della registrazione della pressione arteriosa alfine di favorire una miglior diagnosi da parte delproprio medico curante.
Raccomandiamo di consultare sempre il proprio
medico al fine di interpretare correttamente i dati
relativi alla pressione.

Localizzazione guasti
Problema

Causa probabile

Misura correttiva
Sostituire le batterie usate con
Le batterie sono scariche.
Sul display, nonbatterie nuove.
viene visualizzato
Le batterie sono state inserite Posizionare le batterie in modo talenessun messaggio,
non correttamente.
che i poli positivo e negativo
anche dopo
I poli delle batterie sono
coincidano con quelli indicati nell’accensione.
posizionati in modo errato.
vano batterie.
Il voltaggio delle batterie è
troppo basso. il simbolo(BATTERIE SCARICHE)Il bracciale non si
Sostituire le batterie usate con
lampeggia. In caso di batteriegonfia.
batterie nuove.
completamente scariche, ilsimbolo non saràvisualizzato.
Il bracciale non è avvolto in
Avvolgere il bracciale in modo
modo adeguato.
adeguato.
Durante la fase di
Durante la fase di misurazione,
misurazione, avete mosso ilrimanere fermi e rilassati.
braccio o il corpo.
Il dispositivo nonSedere comodamente e rimanere
esegue lafermi e rilassati. Appoggiare il braccio
misurazione.
La posizione del bracciale
su un tavolo con il palmo della mano
I valori ottenuti sono non è corretta.
rivolto verso l’alto e con il bracciale alla
eccessivamente alti
stessa altezza del cuore.
o bassi.
Nel caso in cui il battito cardiaco
sia molto debole o irregolare, il
dispositivo potrebbe trovaredifficoltà nel determinare lapressione arteriosa.
Italiano

21

Vari

Far riferimento al capitolo “Perché
Il valore al domicilio è diverso
è importante l’automisurazione
da quello misurato in clinica odella pressione arteriosa al propriopresso lo studio medico.
domicilio?”.
Rimuovere le batterie. Posizionarlecorrettamente e eseguirenuovamente la misurazione.

Nota: Nel caso in cui le azioni sopra riportate non risolvano il problema, contattare il rivenditore. Non tentare
di aprire o riparare questo prodotto autonomamente, perché si rischia di invalidare la garanzia.

Manutenzione
Non aprire il dispositivo che include componenti elettrici delicati ed un sofisticato impianto adaria che potrebbero danneggiarsi. Nel caso in cui non si riesce a risolvere il problema anche
dopo aver eseguito quanto riportato nelle istruzioni di cui al capitolo “Localizzazione guasti”,
contattare il centro di assistenza più vicino, oppure rivolgersi al rivenditore o al punto vendita
dal quale ha acquistato il misuratore. Il dispositivo è stato concepito e fabbricato per una
lunga durata di servizio. Tuttavia si raccomanda di controllare lo strumento ogni 2 anni, pergarantire funzionamento e precisione attendibili.

Dati tecnici
Tipo

UA-1020-W

Metodo di misurazione

Oscillometrico

Range di misurazione

Pressione:
0-299 mmHg
Pressione sistolica: 60-279 mmHg
Pressione diastolica: 40-200 mmHg
Pulsazioni:
40-180 battiti/minuto

Precisione

Pressione:
Polso:

Alimentazione

4 batterie da 1,5 V (R6P, LR6 o AA) o
Alimentatore a rete elettrica (TB-233C) (non incluso)

Numero di misurazioni

Circa 1000
LR6 (batterie alcaline)
Circa 300
R6P (batterie al manganese)Con un valore di pressione di 180 mmHg a una
temperatura ambiente di 23 °C

Classificazione

Dispositivo medico elettronico ad alimentazione interna(alimentazione con batterie) /Class II (alimentazione con alimentatore a rete)
Modalità di funzionamento continuo

Test clinici

In conformità ad ISO81060-2：2013±3 mmHg±5 %Nello studio di convalida clinica, K5 è stato utilizzato su 85
soggetti per la determinazione della pressione arteriosa
diastolica.
EMD

IEC 60601-1-2: 2014

Memoria dati

90 misurazioni
Italiano22

Condizioni d’esercizioda +10 a +40 °C / da 15 a 85 % RH / da 800 a 1060 hPa

Condizioni di trasporto /Conservazione
Dimensioni

da -20 a + 60 °C / da 10 a 95 % RH / da 700 a 1060 hPa
Circa 140 (L) x 60 (A) x 105 (P) mm, bracciale escluso

Peso:

Circa 285 g, batterie escluse

Protezione contro l'ingresso
di sostanze

Dispositivo: IP20

Parte applicata
Vita utile

Bracciale Tipo BF
Dispositivo: 5 anni (se utilizzato sei volte al giorno)
Bracciale: 2 anni (se utilizzato sei volte al giorno)

Alimentatore: 5 anni (se utilizzato sei volte al giorno)
Alimentatore opzionale(TB-233C)

L'alimentatore ha lo scopo di collegare il monitor dellapressione sanguigna a una fonte di alimentazione inambiente domiciliare.
Per l’acquisto, contattare il punto vendita autorizzato
A&D.
L'alimentatore a rete elettrica deve essere ispezionato osostituito periodicamente.

Simboli riportati sull’alimentatore
SimboliFunzione / Significato
Per l'uso solo in ambienti interni
Dispositivo di Classe II
Fusibile termico
Fusibile
Etichetta del dispositivo relativa alla
direttiva CE
Etichetta del dispositivo relativa alla
certificazione EAC
Polarità dello spinotto dell'alimentatore
Accessori venduti separatamente
Bracciale
Numero di catalogo
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Dimensione del bracciale
Bracciale grande per adulti
Bracciale ampio
Bracciale per adulti
Bracciale piccolo per adulti

Alimentatore a rete elettrica
Numero di catalogo
TB-233C

Dimensione del braccio
31 cm a 45 cm
22 cm a 42 cm
22 cm a 32 cm
16 cm a 24 cm

Spina
Tipo C

Nota: Le specifiche sono soggette a variazione, senza alcun preavviso.
La classificazione IP indica i gradi di protezione degli involucri in conformità con lanorma CEI 60529. Questo dispositivo è protetto contro corpi solidi estranei di 12 mm
di diametro o più grandi, come per esempio un dito. Questo dispositivo non è
protetto contro i getti d’acqua.
Italiano

23

MEMO

Italiano

24

Inhoud
Beste klant ............................................................................................ 2
Inleidende opmerkingen........................................................................ 2
Voorzorgsmaatregelen .......................................................................... 2
Identificatie onderdelen ........................................................................ 4
Symbolen .............................................................................................. 5
Gebruiksmodus ..................................................................................... 7
De bloeddrukmonitor gebruiken ........................................................... 8
De batterijen plaatsen / vervangen .................................................. 8
De bloeddrukmonitor gebruiken ........................................................... 9
De luchtslang aansluiten ................................................................. 9
De netstroomadapter aansluiten ...................................................... 9
De ingebouwde klok instellen ........................................................ 10
De juiste manchetgrootte selecteren ............................................. 11
De armmanchet aanbrengen .......................................................... 11
Nauwkeurige metingen uitvoeren .................................................. 13
Meting ........................................................................................... 13
Na meting ..................................................................................... 13
Metingen ............................................................................................. 14
Normale meting............................................................................. 14
Meting met de met SET ingestelde druk ......................................... 15
Meting met de gewenste systolische druk...................................... 15
Een nauwkeurige meting uitvoeren ................................................ 16
TriCheckTM-meting.............................................................................. 16
De TriCheckTM-modus selecteren .................................................. 16
Meting met de TriCheckTM-modus ................................................. 17
De gegevens uit het geheugen opvragen ............................................... 18
Waarvoor dient de IHB/AFib-indicator? .................................................. 19
Wat is de AFib? .................................................................................... 19
Drukbalkindicator ............................................................................... 19
Indicator WHO-classificatie ................................................................. 19
Meer over bloeddruk ........................................................................... 20
Wat is bloeddruk? .......................................................................... 20
Wat is hoge bloeddruk en hoe wordt deze onder controle gehouden? 20
Waarom thuis bloeddruk meten? ................................................... 20
Bloeddrukclassificatie WHO ........................................................... 20
Bloeddrukvariaties ......................................................................... 20
Probleemoplossing .............................................................................. 21
Onderhoud.......................................................................................... 22
Technische gegevens .......................................................................... 22

Nederlands 1

Beste klant
Gefeliciteerd met uw aanschaf van een hypermoderne A&D-bloeddrukmonitor.
Deze bloeddrukmonitor, ontworpen voor gebruiksgemak en nauwkeurigheid,
zal uw dagelijkse bloeddrukmetingen vergemakkelijken.
We bevelen u aan deze handleiding zorgvuldig door te lezen voordat u het
apparaat voor het eerst gebruikt.

Inleidende opmerkingenDit apparaat voldoet aan Europese Richtlijn 93/42/EEG betreffende medischehulpmiddelen. Dit wordt aangegeven met de conformiteitsmarkering.
(0123: het referentienummer voor de betrokken aangemelde instantie)
Het apparaat is ontworpen voor gebruik bij volwassenen, niet bij baby’s of peuters.
Gebruiksomgeving. Het apparaat is bedoeld voor gebruik door uzelf voor
medische zorg in de thuissituatie.
Dit apparaat is ontworpen voor het meten van bloeddruk en hartslag vanpersonen voor diagnosedoeleinden.

Voorzorgsmaatregelen
Voor de constructie van dit apparaat zijn precisieonderdelen gebruikt. Vermijd
extreme temperatuur, luchtvochtigheid, direct zonlicht, schokken of stof.
Reinig het apparaat en de manchet met een droge, zachte doek of een doek
die is bevochtigd met water en een neutraal schoonmaakmiddel. Gebruik in
geen geval alcohol, benzeen, thinner of andere agressieve chemische
middelen om het apparaat en de manchet te reinigen.
Vouw de manchet niet te strak op en bewaar de slang niet gedurende langere
perioden in strak opgewikkelde vorm, aangezien dit de levensduur van deonderdelen kan verkorten.
Pas op dat baby’s of peuters zichzelf niet per ongeluk wurgen met de slang en
kabel.
Draai de luchtslang niet tijdens het meten. Dit zou letsel kunnen veroorzaken
als gevolg van continue druk van de manchet.
Het apparaat en de manchet zijn niet waterbestendig. Voorkom dat het
apparaat en de manchet in contact komen met regen, zweet en water.
Metingen kunnen verkeerde waarden aangeven als het apparaat wordt
gebruikt in de nabijheid van televisies, magnetrons, mobiele telefoons,röntgenapparaten of andere apparaten met een sterk elektrisch veld.
Draadloze communicatieapparaten zoals thuisnetwerkapparaten, mobiele
telefoons, draadloze telefoons en de basisstations daarvan en walkietalkies
kunnen van invloed zijn op de bloeddrukmonitor.
Hanteer daarom een minimumafstand van 30 cm van dergelijke apparaten.
Gebruikte apparatuur, onderdelen en batterijen mogen niet als normaal
huishoudelijk afval worden behandeld en moeten worden afgevoerd volgens detoepasselijke plaatselijke regels.
Wanneer u de netstroomadapter gebruikt, dient u ervoor te zorgen dat deadapter snel uit het stopcontact kan worden gehaald indien nodig.
Wanneer u het apparaat opnieuw gebruikt, dient u eerst te controleren of het
Nederlands 2

schoon is.
Breng geen wijzigingen aan in het apparaat. Dit kan ongevallen of schade aan
het apparaat veroorzaken.
Als u de bloeddruk wilt meten, moet de arm hard genoeg worden afgekneld
door de manchet dat de bloedstroom door de slagader tijdelijk wordt gestopt.
Dit kan pijn, gevoelloosheid of een tijdelijke rode verkleuring van de arm
opleveren. Dit doet zich met name voor wanneer de meting meerdere keren inkorte tijd wordt herhaald.
Eventuele pijn, gevoelloosheid of rode verkleuringen verdwijnen na verloop van
tijd.
Het te vaak meten van de bloeddruk kan kwalijk zijn doordat de bloedcirculatie
wordt verstoord. Controleer dat de werking van het apparaat niet resulteert inlangdurige verstoring van de bloedcirculatie wanneer u het apparaat
herhaaldelijk gebruikt.
Als u een borstamputatie hebt ondergaan, dient u een arts te raadplegenvoordat u het apparaat gebruikt.
Zorg ervoor dat kinderen het apparaat niet zonder toezicht gebruiken en
gebruik het apparaat niet binnen bereik van baby’s of peuters. Dit kan
ongevallen of schade veroorzaken.
Het apparaat bevat kleine onderdelen die een risico op verstikking opleveren
indien ze door baby’s of peuters worden ingeslikt.
Koppel de netstroomadapter los wanneer deze niet in gebruik is tijdens het meten.
Het gebruik van accessoires die niet in deze handleiding worden vermeld, kan
een veiligheidsrisico opleveren.
Indien de batterij kortsluiting maakt, kan deze heet worden, met mogelijk
brandwonden tot gevolg.
Laat het apparaat (ongeveer een uur) op kamertemperatuur komen voordat u
het gebruikt.
Er zijn geen klinische tests uitgevoerd bij pasgeborenen en zwangere vrouwen.
Gebruik het apparaat niet bij pasgeborenen of zwangere vrouwen.
Zorg ervoor dat u de batterijen, de DC-ingang en de patiënt niet tegelijkertijd
aanraakt. Dit zou een elektrische schok kunnen veroorzaken.
Pomp de manchet niet op voordat u deze rond de bovenarm hebt gewikkeld.

Contra-indicaties
Hierna volgen voorzorgsmaatregelen voor het juiste gebruik van het apparaat.
Breng de manchet niet aan rond een arm waar ook andere medischeelektronica zijn aangesloten. Het apparaat werkt dan mogelijk niet goed.
Mensen die ernstige circulatieproblemen hebben in de arm, moeten eerst een
arts raadplegen voordat ze het apparaat gebruiken. Dit om medische
problemen te voorkomen.
Voer geen zelfdiagnose uit aan de hand van de meetwaarden en begin niet zelf
een behandeling. Raadpleeg altijd eerst uw arts voor beoordeling en behandeling.
Breng de manchet niet aan rond een arm met een nog niet genezen wond.
Breng de manchet niet aan rond een arm die voor een infuus of bloedtransfusie
wordt gebruikt. Dit kan letsel of ongevallen veroorzaken.
Gebruik het apparaat niet op een plaats waar brandbare gassen aanwezig zijn,
zoals narcosegassen. Dit zou een explosie kunnen veroorzaken.
Gebruik het apparaat niet in omgevingen met een hoge zuurstofconcentratie,
zoals een zuurstofkamer met hoge druk of een zuurstoftent. Dit kan brand of
een explosie veroorzaken.
Nederlands 3

Identificatie onderdelen
DC-ingang
Display

-knop
-knop
SET-knop

Connector luchtaansluiting
START-knop
Aansluitpunt lucht
Luchtslang
Het juiste maatbereik

Slagadermarkering
Armmanchet

Indexmarkering
Batterijklep
Batterijvak

1,5V-batterijen(R6P, LR6 of AA)

Display

GEHEUGEN

Instellingengemiddelde druk

IHB/AFib-symbool
Symbool onjuist aangebrachte manchet

SystolischDruk-

Symbool bewegingsfout
TriCheckTM-symbool

Diastolische
druk

Hartsymbool
Indicator WHO-classificatie
en drukbalkindicator

HartslagBatterij-indicator
AM / PM-markering
Nederlands 4

klokweergave

Symbolen
Symbolen die op de behuizing van het apparaat en de netstroomadapter
zijn gedrukt
Symbolen

SN

IP

Functie / betekenis
Stand-by en Apparaat aanzetten.
Installatie-instructies batterij
Gelijkstroom (DC)
Serienummer
Productiedatum
Type BF: apparaat, manchet enslangen zijn ontworpen om specialebescherming te bieden tegenelektrische schokken.
Label EG-richtlijn medische apparaten
WEEE-label
Fabrikant
EU-vertegenwoordiger
Polariteit van DC-ingang
Internationaal beveiligingssymbool
Droog houden

Aanbevolen handeling

Symbolen die in de display worden weergegeven
Symbolen

Functie / betekenis
Aanbevolen handeling
Wordt weergegeven wanneer een meting De meting wordt
wordt uitgevoerd. Het knippert wanneer uitgevoerd. Blijf zo stilde hartslag wordt waargenomen.
mogelijk zitten.
IHB/AFib-symbool
Wordt weergegeven wanneer een
onregelmatige hartslag wordt
waargenomen.
Het kan gaan branden wanneer een
zeer lichte vibratie, zoals een rilling ofsiddering, wordt waargenomen.
TriCheckTM-modus
Hierin worden drie achtereenvolgende
automatische metingen genomen en
worden de gemiddelde waardes voor
de drie metingen weergegeven.
Wordt weergegeven wanneer een
beweging van lichaam of arm wordt
waargenomen.
Nederlands 5

Er kan een onjuiste
meetwaarde worden
weergegeven. Voer een
nieuwe meting uit.
Blijf stil zitten tijdens het
meten.

Symbolen

Functie / betekenis
Wordt tijdens metingen weergegevenals de manchet te los is aangebracht.

Aanbevolen handeling
Er kan een onjuiste
meetwaarde worden
weergegeven. Breng demanchet correct aan envoer een nieuwe meting uit.

Eerdere metingen die in GEHEUGEN
zijn opgeslagen.
Gemiddelde waarden
BATTERIJ VOL
De batterijstroomindicator tijdens meting.
Vervang alle batterijen door
BATTERIJ BIJNA LEEG
nieuwe exemplaren
De batterij is bijna leeg als dit knippert. wanneer dit symbool
knippert.
Voer de meting opnieuw
uit.
Onstabiele bloeddruk als gevolg van
beweging tijdens meting.
Blijf stil zitten tijdens het
meten.
De systolische en diastolische waarden
liggen binnen 10 mmHg van elkaar.
De drukwaarde is niet toegenomen
Breng de manchet correct
tijdens het opblazen.
aan en voer de meting
De manchet is niet correct
opnieuw uit.
aangebracht.
FOUT HARTSLAGWEERGAVE
De hartslag is niet goed gedetecteerd.
Verwijder de batterijen, druk
op de knop START en zet
de batterijen terug. Neem
Interne fout bloeddrukmonitor
contact op met de verkoper
als de fout nog steeds wordt
weergegeven.
SYS

Systolische bloeddruk in mmHg

DIA

Diastolische bloeddruk in mmHg

PUL

Hartslag per minuut

AM
PM

Gegevens die zijn gemeten tussen
04:00 en 09:59
Gegevens die zijn gemeten tussen
18:00 en 01:59
Drukinstellingen
Geeft de drukwaarde aan die eerderdoor de gebruiker is ingesteld.
Nederlands 6

Gebruiksmodus
1. Normale meting
Druk op de knop START. De bloeddruk wordt gemeten en gegevens worden in
het apparaat opgeslagen. Het apparaat kan de laatste negentig sets gegevens
automatisch in het geheugen opslaan.

2. De gegevens opvragen
Druk op de knopof als u de gegevens in
het geheugen wilt opvragen. Het gemiddeldevan alle metingen wordt weergegeven, zoals
aangegeven in het figuur aan de rechterkant.
Vervolgens worden de geheugengegevens elke-knop wordt gedrukt, als volgt
keer dat op de
weergegeven.
Het gemiddelde van alle ochtendmetingen(AM), uitgevoerd tussen 04:00 en 09:59.
Het gemiddelde van alle avondmetingen(PM), uitgevoerd tussen 18:00 en 01:59.
Meest recente gegevens (No.n, in dit
voorbeeld No.35)
Laatste gegevens (No.1)
Raadpleeg de paragraaf “De geheugengegevens
opvragen” voor meer informatie over het
opvragen van de gegevens. (Zie pagina 18)

3. Alle gegevens in het geheugen wissen
Druk de knoppen
en tegelijkertijd in. Het
symboolen de batterij-indicator worden
weergegeven. Houd de knoppen
en ingedrukt
begint te knipperen om
tot het verlichte symbool
alle gegevens die in het geheugen staan, te wissen.

4. Meting met de gewenste systolische druk
Zie pagina 15 voor metingen met de gewenste systolische druk.

Nederlands 7

De bloeddrukmonitor gebruiken
De batterijen plaatsen / vervangen
1. Verwijder de batterijklep.
2. Verwijder de gebruikte batterijen enplaats nieuwe batterijen in het
batterijvak, zoals hier weergegeven.
Zorg ervoor dat de polariteit (+) en (-)
correct is.
Gebruik uitsluitend R6P-, LR6- of
AA-batterijen.
3. Zet de batterijklep weer op zijn plaats.

Stap 1
Stap 3

Stap 1
Stap 2

LET OP
Plaats de batterijen in het batterijvak zoals weergegeven. Als ze niet goed
worden geplaatst, werkt het apparaat niet.
Als het symbool(BATTERIJ BIJNA LEEG) knippert op de display, dient u
alle batterijen door nieuwe te vervangen. Mix nooit oude en nieuwe batterijen. Dit
kan de levensduur van de batterijen verkorten. Daarnaast kan het ervoor zorgen
dat het apparaat defect raakt.
Vervang de batterijen twee of meer seconden nadat u het apparaat hebt uitgezet.
(BATTERIJ BIJNA LEEG) zelfs na vervanging van deAls het symbool
batterijen wordt weergegeven, dient u een bloeddrukmeting uit te voeren. Het ismogelijk dat het apparaat de nieuwe batterijen dan wel herkent.
Het symbool (BATTERIJ BIJNA LEEG) wordt niet weergegeven als debatterijen helemaal leeg zijn.
De levensduur van de batterijen varieert al naargelang de omgevingstemperatuur
en kan korter zijn bij lage temperaturen. Vier nieuwe LR6-batterijen gaan
doorgaans ongeveer een jaar mee wanneer u het apparaat tweemaal daags voor
metingen gebruikt.
Gebruik alleen de aangegeven batterijen. De batterijen die met het apparaat
worden meegeleverd, zijn bedoeld voor het testen van de bloeddrukmonitor enhebben mogelijk een beperkte levensduur.
Verwijder de batterijen als u het apparaat langere tijd niet gebruikt.
De batterijen kunnen gaan lekken en een defect veroorzaken.

Nederlands 8

De bloeddrukmonitor gebruiken
De luchtslang aansluiten
Plaats de connector van de luchtslang
stevig in het aansluitpunt voor lucht.

Aansluitpunt lucht

Connector luchtaansluiting

De netstroomadapter aansluiten
Plaats de connector voor denetstroomadapter in de DC-ingang.
Sluit de netstroomadapter vervolgens
aan op een stopcontact.

Connector netstroomadapter

DC-ingang

Gebruik de gespecificeerde netstroomadapter.
(Zie pagina 23.)
Wanneer u de netstroomadapter wilt loskoppelen van het stopcontact, moet u dit
doen door de adapter zelf vast te grijpen en deze uit het stopcontact te trekken.
Wanneer u de netstroomadapter wilt loskoppelen van de bloeddrukmonitor,
moet u dit doen door de connector vast te grijpen en deze uit debloeddrukmonitor te trekken.

Nederlands 9

De bloeddrukmonitor gebruiken
De ingebouwde klok instellen
Stel de klok in voordat u het apparaat gebruikt.
1. Druk op de knop SET tot het jaar begint teknipperen.
2. Selecteer het jaar met de knopof .
Druk op de knop SET om het huidige jaar in testellen en ga naar de maand/dag-selectie. U kunteen datum instellen die tussen 2010 en 2059 ligt.

Druk op

Jaar

of
Maand

3. Selecteer de maand met de knopof .
Druk op de knop SET om de huidige maand in
te stellen en ga naar de dagselectie.

Dagof button.
4. Select de dag met de knop
Druk op de knop SET om de huidige dag in testellen een ga naar de uur/minuut-selectie.
5. Selecteer het uur met de knopof .
Druk op de knop SET om het huidige uur in testellen en ga naar de minuut-selectie.
of .
6. Selecteer de minuut met de knop
(Druk op de knop SET om naar dedrukinstelmodus te gaan. Zie pagina 15 voor
meer informatie.)
Druk op de knop START om het apparaat uit tezetten.

of

of

Uur

MinuutOpmerking: als het apparaat drie minuten niet wordt
gebruikt, wordt het automatisch uitgeschakeld.
Als de klik niet is ingesteld,
Uitzetten
worden in de klokdisplay
Als
u
op
de
liggende
streepjes
START-knop drukt,
weergegeven, zoals u rechts
wordt
het apparaat
kunt zien.
altijd uitgezet.
Als u de knop
ofingedrukt
houdt,
verandertdewaardevoortdurend.

Nederlands 10

of

of

Druk
-instelmodus

De bloeddrukmonitor gebruiken
De juiste manchetgrootte selecteren
Voor het verkrijgen van een nauwkeurige meting, is het belangrijk dat u de juiste
manchetgrootte gebruikt. Als de manchet niet de juiste maat heeft, kan de meting
een onjuiste bloeddrukwaarde opleveren.
De armomvang is op elke manchet gedrukt.
De index- en het juiste maatbereik op de manchet laten u zien of u de juiste
manchet gebruikt. (Zie de tabel “Symbolen die op de manchet zijn gedrukt” op devolgende pagina)
Als de index- buiten het maatbereik wijst, dient u contact op te nemen met uw
plaatselijke verkoper om een vervangende manchet te kopen.
De armmanchet is een verbruiksitem. Koop een nieuw exemplaar als de oude
versleten raakt.
Armomvang
Aanbevolen manchetgrootte
Catalogusnummer
31 cm tot 45 cm
Grote manchet volwassenen
CUF-F-LA
22 cm tot 42 cm
Manchet voor breed spectrum
CUF-I
22 cm tot 32 cm
Manchet volwassenen
CUF-F-A
16 cm tot 24 cm
Kleine manchet voor volwassenen
CUF-F-SA
Armomvang: de omtrek van de arm ter hoogte van de biceps.

De armmanchet aanbrengen
1. Wikkel de manchet rond de bovenarm,ongeveer 1 - 2 cm boven debinnenzijde van de elleboog, zoals inde illustratie.
Plaats de manchet direct tegen de huid.
Kleding kan een zwakkere
hartslagmeting veroorzaken en kan zo
een meetfout tot gevolg hebben.

Rol uw mouwen niet te strak op.
Manchet1-2 cmLuchtslang
Slagadermarkering

2. Het afklemmen van de bovenarm
door het te strak oprollen van een
mouw kan voor een onnauwkeurige
meting zorgen.
3. Ga na of de indexmaatbereik valt.

Maak deklittenbandsluiting
stevig vast.

binnen het juiste
Index-

Nederlands 11

en het juiste maatbereikDe bloeddrukmonitor gebruiken
Symbolen die op de manchet zijn gedrukt
Symbolen
Functie / betekenis

Aanbevolen handeling
Plaats de-markering op
deslagadervandebovenarm of in één lijn met
de ringvinger aan debinnenzijde van de arm.

Slagadermarkering

▲
REF

Index
Catalogusnummer
Het juiste maatbereik voor de manchet
voor volwassenen.
Het is gedrukt op de manchet voor
volwassenen.
Gebruik de grote manchet
Groter dan het bereik dat is gedrukt op de voor volwassenen in plaatsmanchet voor volwassenen en de van de manchet voor
volwassenen of de manchet
manchet voor breed spectrum.
voor breed spectrum.
Het juiste maatbereik voor de manchet
voor breed spectrum. Het is gedrukt op demanchet voor breed spectrum.
Het juiste maatbereik voor de grotemanchet voor volwassenen. Het is gedrukt
op de grote manchet voor volwassenen.
Gebruik de kleine manchet
Kleiner dan het bereik dat is gedrukt op de voor volwassenen in plaatsmanchet voor volwassenen en de manchet van de manchet voor
volwassenen of de manchet
voor breed spectrum.
voor breed spectrum.
Het juiste maatbereik voor de kleinemanchet voor volwassenen. Het is op de
kleine manchet voor volwassenen gedrukt.
Gebruik de manchet voorin plaats van
Het is op de grote en kleine manchet voor volwassenendegrotemanchetvoor
volwassenen gedrukt.
volwassenen of de kleinemanchet voor volwassenen.
Partijnummer
Het juiste maatbereik

Grote manchet volwassenen

A

L

Manchet voor breed spectrum

S

W

L

Manchet volwassenen

S

A

L

Kleine manchet voor volwassenen

S

A

Nederlands 12

De bloeddrukmonitor gebruiken
Nauwkeurige metingen uitvoeren
Ga als volgt te werk voor de meest nauwkeurige bloeddrukmeting:
Ga comfortabel op een stoel zitten. Laat uw arm op de tafel rusten. Zorg ervoor
dat u uw benen niet kruist. Houd uw voeten plat op de vloer en houd uw rug recht.
Ontspan gedurende vijf tot tien minuten voordat u gaat meten.
Plaats het midden van de manchet ter hoogte van uw hart.
Blijf stil zitten en praat niet tijdens het meten.
Voer de meting niet uit als u zich net fysiek hebt ingespannen of een bad hebt
genomen. Rust ongeveer twintig tot dertig minuten voordat u de meting uitvoert.
Probeer uw bloeddruk iedere dag rond dezelfde tijd te meten.

MetingHet is normaal als de manchet erg knelt tijdens het meten. (Maak u niet ongerust).

Na meting
Druk na meting op de knop START om het apparaat uit te zetten.
Verwijder de manchet en registreer uw gegevens. Na één minuut inactiviteit wordt
het apparaat automatisch uitgezet.

Nederlands 13

Metingen
Lees voordat u gaat meten de paragraaf “Een nauwkeurige meting uitvoeren”
op pagina 16.

Normale meting
1.

Plaats de manchet rond de arm (bij voorkeur
Ter hoogte van het hart
de linkerarm).
Zit stil tijdens het meten.

2.

Druk op de knop START .
Alle displayonderdelen worden weergegeven.
Nul (0) wordt weergegeven en knippert een
korte tijd.
De weergave op de display verandert, zoals
te zien op de afbeelding rechts, op het
moment dat de meting begint. De manchet
wordt opgeblazen. Het is normaal als demanchet erg knelt. Tijdens het opblazen
wordt
links
op
de
displayeen
drukbalkindicator weergegeven.
Opmerking: als u het opblazen op enig
moment wilt stoppen, drukt uopnieuw op de knop START .

3.

4.

5.

Zodra de manchet volledig is opgeblazen,
begint het leeglopen automatisch. Het
hartsymbool
gaat knipperen, wat
aangeeft dat er wordt gemeten. Zodra dehartslag wordt gedetecteerd, knippert het
symbool bij elke hartslag.
Opmerking: als er geen passende druk wordt
bereikt, begint het apparaat
automatisch opnieuw met opblazen.
Raadpleeg de paragraaf “Meting met
de met SET ingestelde druk” op devolgende pagina als u opnieuw
opblazen wilt vermijden.
Zodra de meting voltooid is, worden desystolische en diastolische drukwaarden en dehartslag weergegeven.
De manchet laat de resterende lucht
ontsnappen en loopt volledig leeg.
Druk op de knop START om het apparaat
uit te zetten.
Na één minuut inactiviteit wordt het
apparaat automatisch uitgezet.

Opmerking: zorg voor een tussenpauze van
minstensdrie
minutenvoor
metingen bij dezelfde persoon.
Nederlands 14

Druk op

START

Alle displayonderdelen in beeld

Nul-display
Start het opblazen

Bezig met
opvoeren druk

De meting wordt
uitgevoerd

Systolische druk
Diastolische druk
WHO-classificatie
HartslagResterende lucht wordt
automatisch uitgestoten

Metingen
De UA-1020-W is ontworpen om de hartslag te detecteren en de manchet
automatisch tot een systolisch drukniveau op te blazen.
Als herhaaldelijk opnieuw wordt opgeblazen, dient u de volgende methoden te
hanteren.

Meting met de met SET ingestelde drukHet kan tijdens de bloeddrukmeting voorkomen dat de manchet opnieuw wordt
opgeblazen.
Er kan een vaste drukwaarde worden ingesteld om opnieuw opblazen tevoorkomen.

1.

2.

Druk bij stap 6 van de klokinstelprocedure (ziepagina 10), op de knop SET om naar de
druk-instelmodus te gaan. De waarde die
momenteel is ingesteld, knippert.

ofDruk op de knopofom een drukwaarde
te selecteren die 30 mmHg of meer boven uw
verwachte systolische druk ligt uit de volgende
opties.
AUTO : Automatisch opblazen (standaardwaarde)
180 : Drukwaarde van 180 mmHg (vast)
210 : Drukwaarde van 210 mmHg (vast)
Uitzetten
240 : Drukwaarde van 240 mmHg (vast)

3.

TriCheck-instelmodus

Druk op de knop SET om naar de TriCheck-instelmodus te gaan.
Druk op de knop START om het apparaat uit te zetten. Als het apparaat drieminuten niet wordt gebruikt, wordt het automatisch uitgeschakeld.
De volgende meting wordt uitgevoerd met de nieuwe drukwaarde.

Meting met de gewenste systolische druk

Gebruik de volgende methode wanneer er herhaaldelijk opnieuw wordt
opgeblazen terwijl de drukwaarde via de voorgaande procedure op 240 isingesteld of wanneer de resultaten niet worden weergegeven, zelfs als de druk
tot 20 mmHg of lager daalt.

1.

Plaats de manchet rond de arm (bij voorkeur delinkerarm).

2.

Houd de knop START ingedrukt terwijl denul knippert, tot een getal wordt
weergegeven dat ongeveer 30 tot 40
mmHg hoger is dan uw verwachtesystolische druk.

3.

Zodra het gewenste getal is bereikt, laatu de knop START los om de meting testarten. Ga door met het meten van uw
bloeddruk zoals beschreven op de vorige
pagina.

Nederlands 15

Ter hoogte van
het hart

Houd de knop ingedrukt

START

Laat de knop los bij degewenste systolischedruk

Zie de vorige pagina
voor meting

Een nauwkeurige meting uitvoeren

Ga in een comfortabele houding zitten. Plaats uw arm op een tafel met dehandpalm omhoog en de manchet ter hoogte van uw hart.
Ontspan gedurende vijf tot tien minuten voordat u gaat meten. Als u opgewondenof depressief bent als gevolg van emotionele stress, zal de meting deze stressweerspiegelen in de vorm van een hogere (of lagere) bloeddrukwaarde dannormaal en zal de hartslag meestal sneller zijn dan normaal.

De bloeddruk van een persoon varieert constant, afhankelijk van wat u doet en
wat u hebt gegeten. Wat u drinkt kan een zeer sterk en snel effect hebben op
uw bloeddruk.
Dit apparaat baseert zijn metingen op de hartslag. Als u een erg zwakke of
onregelmatige hartslag hebt, kan het apparaat problemen hebben bij het
bepalen van uw bloeddruk.
Als het apparaat een toestand aantreft die abnormaal is, stopt het met meten
en geeft het een foutsymbool weer. Zie pagina 6 voor een beschrijving van desymbolen.
Deze bloeddrukmonitor is bedoeld voor gebruik door volwassenen. Raadpleeg
uw arts voordat u dit apparaat op een kind gebruikt. Kinderen hebben toezicht
nodig bij het gebruik van dit apparaat.
De prestatie van de automatische bloeddrukmonitor kan worden beïnvloeddoor buitensporige temperaturen, luchtvochtigheid of hoogte.

TriCheckTM-meting
In de TriCheckTM-modus worden automatisch drie achtereenvolgendemetingen
uitgevoerd en worden de gemiddelde waarden van de drie metingen
weergegeven.

De TriCheckTM-modus selecteren
1.

Druk in de druk-instelmodus op de knop
SET als u de TriCheckTM-instelmodus
wilt activeren. De waarde die
momenteel is ingesteld, knippert.

2.

of
om de
Druk op de knop
TM
TriCheck -modus AAN (ON) of UIT(OFF) te zetten.
ON: TriCheckTM-modus
OFF: Normale meetmodus (standaardwaarde)

3.

Druk op de knop START of SET om het
apparaat uit te zetten. Als het apparaat
drie minuten niet wordt gebruikt, wordt
het automatisch uitgeschakeld.

In de druk-instelmodus:
druk op

De waarde die momenteel isingesteld, knippert

ofWijzig de instellingof
Nederlands 16

START

Meting met de TriCheckTM-modus
1.

2.

Druk op de knop START .
Alle
displayonderdelen
wordenweergegeven.
Nul (0) wordt weergegeven en knippert
een korte tijd. Daarna start de eerste
meting.
Zodra de meting voltooid is, worden desystolische en diastolische drukwaarden
endehartslagweergegeven.
Vervolgens start een countdown van
één minuut tot de tweede meting.

3.

Na één minuut start de tweede meting.

4.

Zodra de meting voltooid is, worden degemiddelde waarden van de eerste en
tweedemetingweergegeven.
Vervolgens start een countdown van
één minuut tot de derde meting.

5.

Na één minuut start de derde meting.

6.

Zodra de meting voltooid is, worden degemiddelde waarden van de driemetingen weergegeven en worden deze
in het geheugen opgeslagen.

Opmerkingen:
Tijdens het meten wordt het
TriCheckTM-symbool weergegeven.
Druk op de knop START als u demeting wilt stoppen. Er worden in dit
geval geen gegevens in het
geheugen opgeslagen.
Indien tijdens de eerste meting het
symboolvooreen
onjuistaangebrachtemanchetwordt
weergegeven, dient u de meting teannuleren, de manchet op de juistewijze aan te brengen en een nieuwe
meting te starten.
Nadat de metingen voltooid zijn,
worden de gemiddelde waarden voor
de drie metingen in het geheugen
opgeslagen.
Er worden geen gegevens opgeslagenindien u de metingen annuleert voordat
de derde meting is voltooid.
Nederlands 17

Ter hoogte van het hart

START

Druk opAlle displayonderdelen in beeld
Eerste meting

Een minuut later

Resultaten van deeerste meting

Tweede meting

Gemiddelde van deeerste en tweede meting

Een minuut later
Derde meting

Gemiddelde van dedrie metingenDe gegevens uit het geheugen opvragen
Opmerking: het apparaat slaat de laatste negentig gegevenssets automatisch op in
het geheugen.
1. Druk op de knopof .
Druk opHet gemiddelde van alle metingen en
het aantal gegevenssets worden
weergegeven.
(Als er geen gegevens zijn, wordt “0”weergegeven. Druk op de knop , of
START om het apparaat uit te zetten.)
2. Iedere keer dat op de knop
wordt
gedrukt (of op de knop
, om gegevensin omgekeerde volgorde weer te geven),
worden de geheugengegevens als volgt
weergegeven.
Het
gemiddeldevanalleochtendmetingen (AM), uitgevoerd
tussen 04:00 en 09:59.
(In het voorbeeld, 10 metingen. Als ergeen gegevens zijn, wordt “--”weergegeven.)Het gemiddelde van alle avondmetingen(PM), uitgevoerd tussen 18:00 en 01:59.
(In het voorbeeld, 9 metingen. Als ergeen gegevens zijn, wordt “--”weergegeven.)
Meest recente gegevens (No.n, in dit
voorbeeld No.35)
Drie seconden na de weergave van
het gegevensnummer, worden demeetgegevens weergegeven.

ofGemiddelde systolische druk
Gemiddelde diastolische druk
Gemiddelde hartslag

Gemiddelde vanalle gegevens

Gemiddelde systolische druk
Gemiddelde diastolische druk
Gemiddelde hartslag
Gemiddelde vande AM-gegevens

Gemiddelde systolische druk
Gemiddelde diastolische druk
Gemiddelde hartslag

Gemiddelde vande PM-gegevens

SystolischDiastolischHartslagMeest recente gegevens

Laatste gegevens (No.1)
Drie seconden na de weergave van het
gegevensnummer, worden demeetgegevens weergegeven.
3. Druk nadat de laatste gegevens zijn
weergegeven op de knop
om opnieuw
het gemiddelde van alle metingen weer tegeven.
4. Druk op de knop START om het apparaat
uit te zetten. Na één minuut inactiviteit
wordt het apparaat automatisch uitgezet.
Nederlands 18

SystolischDiastolischHartslagLaatste gegevens (oudste)

Waarvoor dient de IHB/AFib-indicator?
Zodra de bloeddrukmonitor tijdens de metingen een onregelmatige hartslag
waarneemt, wordt de IHB/AFib-indicator op de display weergegeven naast de
meetwaarden.
Opmerking: we bevelen aan dat u contact met uw huisarts opneemt als u deze
IHB/AFib-indicator regelmatig ziet.

Wat is de AFib?
Het hart trekt samen als gevolg van elektrische signalen in het hart en stuurt bloeddoor het lichaam. Atriumfibrillatie (AFib) doet zich voor wanneer het elektrische
signaal in het atrium verward raakt, wat leidt tot storingen in de hartslaginterval.
AFib kan ervoor zorgen dat bloed in het hart stagneert, wat makkelijk kan leiden
tot bloedproppen, een oorzaak voor beroertes en hartaanvallen.

Drukbalkindicator
De indicator bewaakt de drukontwikkeling tijdens metingen.
Bezig met opvoeren druk

Bezig met leeglopen

Bezig met opblazen

Bezig met meten
Opblazen is voltooid

Indicator WHO-classificatie
Elk segment van de balkindicator komt overeen
met de bloeddrukclassificatie van de WHO die
op de volgende pagina wordt beschreven.

Indicator WHO-classificatie
Ernstige hoge bloeddruk
Matige hoge bloeddruk
Milde hoge bloeddruk
Hoog-normaal
NormaalOptimaal

Voorbeeld:

Matige hoge bloeddruk

Milde hoge bloeddruk: De indicator geeft een segment weer op
basis van de huidige gegevens,corresponderend met de WHO-classificatie.

Hoog-normaal
Nederlands 19

Meer over bloeddruk
Wat is bloeddruk?

Bloeddruk is de kracht die door het bloed wordt uitgeoefend tegen de wanden
van de slagaders. Systolische druk doet zich voor wanneer het hart samentrekt.
Diastolische druk doet zich voor wanneer het hart uitzet. Bloeddruk wordt
gemeten in millimeter kwik (mmHg). Uw natuurlijke bloeddruk wordt
vertegenwoordigd door de fundamentele druk, die ‘s ochtends wordt gemeten
terwijl u nog in ruste bent en voordat u ontbijt.

Wat is hoge bloeddruk en hoe wordt deze onder controle gehouden?

Hoge bloeddruk is een abnormaal hoge slagaderlijke bloeddruk die, indien
onbehandeld, tot een groot aantal gezondheidsproblemen kan leiden,
waaronder beroertes en hartaanvallen. Hoge bloeddruk kan onder controle
worden gehouden door het veranderen van uw levensstijl, het vermijden vanstress en met medicatie onder dokterstoezicht.
U kunt het volgende doen om hoge bloeddruk te voorkomen of onder controle
te houden:
Niet roken
Regelmatig bewegen
Minder zout en vet in uw voedsel
Regelmatig een medische controle
gebruiken
ondergaan
Een gezond gewicht handhaven

Waarom thuis bloeddruk meten?

Als de bloeddruk in het ziekenhuis of bij een huisarts wordt gemeten, kan dit
spanning veroorzaken, wat tot een verhoogde waarde kan leiden, 25 tot 30 mmHg
hoger dan wanneer thuis gemeten. Thuismeting reduceert de effecten van externe
invloeden op de bloeddrukwaarden, geeft een aanvulling op de metingen van dearts en leidt tot een nauwkeuriger en meer complete bloeddrukgeschiedenis.

Bloeddrukclassificatie WHO

Bloeddrukvariaties

Diastolische bloeddruk

De Wereldgezondheids- organisatie (WHO) heeft normen opgesteld voor het
beoordelen van bloeddruk, Referentiemateriaal: Journal of Hypertension 1999, Vol 17, nr. 2
zonder onderscheid te maken mmHg
Hoge bloeddruk type 3 (ernstig)
110
naar leeftijd. Deze normen
Hoge bloeddruk type 2 (gemiddeld)105
worden weergegeven in de100
grafiek.
95
90
85
80

Hoge bloeddruk type 1 (mild)

Hoog-normaal
De bloeddruk van een persoon
Normaalkan flink variëren, zowel in de
Optimaal
loop van de dag als tijdens het
120 130 140 150 160 170 180
seizoen. Er kan een verschil van
Systolische bloeddruk
mmHg
30 tot 50 mmHg zijn als gevolg
van diverse omstandigheden in de loop van de dag. Bij personen met hoge
bloeddruk zijn deze variaties zelfs nog groter. Normaal gesproken stijgt debloeddruk terwijl u werkt of aan sport doet en is deze op zijn laagste punt wanneer
u slaapt. Maak u daarom niet al te ongerust over de resultaten van één enkele
meting.

Nederlands 20

Normale fluctuatie binnen een dag(om de vijf minuten gemeten)
SystolischBloeddrukDoe uw metingen elke dag op dezelfde tijd en
gebruik daarvoor de procedure die in deze
handleiding wordt beschreven om uw normale
bloeddruk te achterhalen. Regelmatige metingen
geven
een
meer
volledigebloeddrukgeschiedenis. Zorg ervoor dat u ook dedatum en tijd opschrijft wanneer u uw bloeddruk
noteert. Raadpleeg uw arts om uw
bloeddrukgegevens te interpreteren.

Slaap

Diastolisch

PM

AM

Tijd

Probleemoplossing
Probleem
Er wordt niets op
de display
weergegeven,terwijl het
apparaat aan
staat.

Mogelijke oorzaak

Aanbevolen handeling
Vervang alle batterijen door nieuwe
De batterijen zijn leeg.
exemplaren.
Plaats de batterijen opnieuw en
De contactpunten van dezorg ervoor dat de positieve
batterijen zitten niet op decontactpunten op de posities zitten
juiste positie.
die in het batterijvak zijn aangeduid.
Het batterijvoltage is telaag.
Het symbool
De manchet(BATTERIJ BIJNA LEEG) Vervang alle batterijen door
wordt niet
knippert. Als de batterijen nieuwe exemplaren.
opgeblazen.
helemaal leeg zijn, wordt
het symbool niet
weergegeven.
De manchet is niet correct
Breng de manchet correct aan.
aangebracht.
U hebt uw arm of lichaam Zorg ervoor dat u stil blijft zitten en
bewogen tijdens het
geen geluid maakt tijdens
meten.
metingen.
Het apparaat
voert geen
Zorg ervoor dat u comfortabel enmeting uit.
stil zit.
De waarden zijn De positie van dePlaats uw arm op een tafel met demanchetisnietcorrect.
te hoog of te
handpalm omhoog en de manchetlaag.
ter hoogte van uw hart.
Als u een erg zwakke of
onregelmatige hartslag hebt, kan
het apparaat problemen hebben
bij het bepalen van uw bloeddruk.
De waarde wijkt af van dewaarde die bij een bezoek Zie “Waarom thuis bloeddruk
meten?”.
aan een ziekenhuis of
huisarts is gemeten.
OverigVerwijder de batterijen. Plaats ze
op de goede manier terug en voer
nog een meting uit.
Opmerking: neem contact op met de verkoper als u het probleem niet kunt oplossen
met de hierboven beschreven handelingen. Probeer dit product niet te
openen of repareren. Als u dit toch doet, wordt uw garantie ongeldig.
Nederlands 21

OnderhoudOpen het apparaat niet. Er worden in dit apparaat delicate onderdelen en een
ingewikkelde luchteenheid gebruikt die beschadigd zouden kunnen raken. Alsu het probleem niet kunt oplossen aan de hand van deprobleemoplossingsinstructies, dient u contact op te nemen met degeautoriseerde verkoper in uw gebied of met onze afdeling klantenservice. De
klantenservice van A&D voorziet geautoriseerde verkopers van technische
informatie, reserveonderdelen en reserve-eenheden.
Het apparaat is ontworpen en gefabriceerd voor een lange levensduur. Wij
bevelen echter over het algemeen aan het apparaat elke 2 jaar te laten
inspecteren om te controleren dat dit nog goed werkt en nauwkeurig is. Neem voor
onderhoud contact op met de geautoriseerde verkoper in uw gebied of met A&D.

Technische gegevens
Type
Meetmethode
Meetbereik

Meetnauwkeurigheid
Voeding
Aantal metingen

Classificatie

Klinische test

EMD
GeheugenGebruiksomstandigheden
Omstandigheden voortransport / opslag
Afmetingen
Gewicht

UA-1020-W
Oscillometrische metingDruk:
0 - 299 mmHg
Systolische druk: 60 - 279 mmHg
Diastolische druk: 40 - 200 mmHg
Hartslag: 40 - 180 slagen / minuutDruk:± 3 mmHg
Hartslag: ± 5 %4 x 1,5 V-batterijen (R6P, LR6 of AA) of
netstroomadapter (TB-233C) (Niet inbegrepen)
Ongeveer 1.000 keer LR6 (alkalinebatterijen)
Ongeveer 300 keer R6P (mangaanbatterijen)
Met een drukwaarde van 180 mmHg bij een
kamertemperatuur van 23°C.
ME-apparatuur met interne voeding (voeding via
batterijen) /Klasse II (voeding via adapter)
Modus continu bedrijf
Volgens ISO81060-2：2013
Tijdens de klinische validatiestudie werd K5 op 85
proefpersonen gebruikt om de diastolische bloeddruk
te bepalen.
IEC 60601-1-2: 2014
Laatste 90 metingen+10 tot +40 °C / relatieve luchtvochtigheid 15 tot 85% /800 tot 1.060 hPa-20 tot +60 °C / relatieve luchtvochtigheid 10 tot 95% /700 tot 1.060 hPa
Ongeveer 140 [B] x 60 [H] x 105 [D] mm
Ongeveer Ongeveer 285 g, exclusief batterijen
Nederlands 22

Aangebracht onderdeel
Beschermingsgraad
Nuttige levensduur

Manchet Type BF
Apparaat: IP20
Apparaat: 5 jaar (indien zes keer per dag gebruikt)
Manchet: 2 jaar (indien zes keer per dag gebruikt)
Netstroomadapter: 5 jaar (indien zes keer per dag
gebruikt)

Accessoire
netstroomadapter
TB-233C

De adapter wordt gebruikt om de bloeddrukmonitor thuis
op een voedingsbron aan te sluiten.
Neem voor aankoop contact op met uw plaatselijke
A&D-verkoper.
De netstroomadapter moet periodiek wordengeïnspecteerd of vervangen.
Symbolen die op de netstroomadapter zijn gedrukt
Symbolen
Functie / betekenis
Uitsluitend voor gebruik binnenshuis
Klasse II-apparaat
Thermische zekering
Zekering
Apparaatlabel EG-richtlijn
Apparaatlabel EAC-certificatie
Polariteit connector netstroomadapter

Accessoires worden afzonderlijk verkocht
ManchetCatalogusnummer
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Manchetgrootte
Grote manchet
volwassenen
Manchet voor breed
spectrum
Manchetvolwassenen
Kleine manchet
voor volwassenen

Armomvang
31 cm tot 45 cm
22 cm tot 42 cm
22 cm tot 32 cm
16 cm tot 24 cm

Netstroomadapter
Catalogusnummer
TB-233C

Stekker
Type C

Opmerking: specificaties kunnen zonder voorafgaande kennisgeving worden
gewijzigd.
De IP-classificatie is de mate van bescherming die door behuizingen
wordt geboden conform IEC 60529. Dit apparaat is beschermd tegenvaste vreemde voorwerpen met een diameter van 12 mm of groter,zoals vingers. Dit apparaat is niet beschermd tegen water.

Nederlands 23

MEMO

Nederlands 24

Innehållsförteckning
Bästa kund ............................................................................................ 2
Inledande anmärkningar ....................................................................... 2
Försiktighetsåtgärder ............................................................................ 2Ingående delar ...................................................................................... 4
Symboler ............................................................................................... 5
Driftsläge .............................................................................................. 7
Använda mätaren .................................................................................. 8
Installera/byta batterier ................................................................... 8
Ansluta luftslangen ......................................................................... 9
Ansluta nätadaptern ........................................................................ 9
Ställa in den inbyggda klockan ...................................................... 10
Välja rätt manschettstorlek ............................................................ 11
Sätta på armmanschetten .............................................................. 11Så här görs korrekta mätningar ..................................................... 13
Mätning ......................................................................................... 13
Efter mätningen ............................................................................. 13
Mätningar ........................................................................................... 14
Normal mätning ............................................................................ 14
Mäta med SET-trycket ................................................................... 15
Mäta med önskat systoliskt tryck ................................................... 15
Anmärkningar för korrekt mätning ................................................ 16
TriCheckTM-mätning............................................................................ 16
Välj TriCheckTM-läget .................................................................... 16
Mäta i TriCheckTM-läget ................................................................. 17
Hämta minnesdata .............................................................................. 18
Vad är IHB/AFib-indikatorn?................................................................ 19
Vad är AFib (förmaksflimmer)? ............................................................ 19
Tryckindikator ..................................................................................... 19
Indikator för WHO-klassificering ......................................................... 19
Om blodtryck ...................................................................................... 20
Vad är blodtryck?........................................................................... 20
Var är hypertoni och hur kontrolleras det? ..................................... 20
Varför mäta blodtrycket i hemmet? ................................................ 20
WHO:s blodtrycksklassificering ...................................................... 20
Variationer i blodtrycket ................................................................ 20
Felsökning .......................................................................................... 21
Underhåll ............................................................................................ 22
Tekniska data...................................................................................... 22

Svenska 1

Bästa kund
Gratulerar till ditt inköp av en toppmodern blodtrycksmätare från A&D.
Mätaren är utformad för att vara enkel att använda och samtidigt vara
noggrann och underlättar på så sätt din dagliga blodtrycksmätningsrutin.
Vi rekommenderar att du läser igenom bruksanvisningen noggrant
innan du använder enheten för första gången.

Inledande anmärkningar
Enheten överensstämmer med Rådets direktiv 93/42/EEG om-symbolen för
medicintekniska produkter. Det indikeras genomöverensstämmelse.
(0123: Referensnumret avser det aktuella anmälda organet)
Enheten är utformad för användning på vuxna, inte på nyfödda ellersmåbarn.
Användningsmiljö. Enheten är avsedd för personligt bruk för vård i hemmet.
Enheten är avsedd för att mäta blodtrycket och pulsfrekvensen hos
människor i diagnostiskt syfte.

Försiktighetsåtgärder
Precisionskomponenter har använts vid tillverkningen av enheten. Undvik
extrema temperaturer, luftfuktighet, direkt solljus, stötar och damm.
Rengör enheten och manschetten med en torr, mjuk trasa eller en trasa
fuktad med vatten och ett neutralt rengöringsmedel. Använd aldrig
alkohol, bensen, thinner eller andra aggressiva kemikalier för att rengöra
enheten eller manschetten.
Undvik att vika ihop manschetten för hårt och förvara inte slangen hårtlindad under längre tidsperioder eftersom det kan förkorta
komponenternas livslängd.
Vidta åtgärder så att det inte finns risk för att spädbarn och små barn
stryps av slangen och kabeln.
Vrid inte luftslangen under mätningen. Det kan orsaka personskador på
grund av det kontinuerliga manschettrycket.
Enheten och manschetten är inte vattentäta. Undvik regn, svett ochvatten på enheten och manschetten.
Mätningarna kan bli felaktiga om enheten används i närheten av
TV-apparater, mikrovågsugnar, mobiltelefoner eller röntgenutrustningeller andra enheter med kraftiga magnetfält.
Trådlösa kommunikationsenheter, t.ex. enheter i hemmanätverk,
mobiltelefoner, trådlösa telefoner och deras basstationer samt
walkie-talkies kan påverka blodtrycksmätaren.
Därför ska sådana enheter hållas på minst 30 cm avstånd.
Använd utrustning, delar och batterier betraktas inte som vanligthushållsavfall och måste kasseras i enlighet med lokala föreskrifter.
När nätadaptern används, kontrollera att nätadaptern enkelt kan dras ur
Svenska 2

eluttaget vid behov.
Kontrollera att enheten är ren när den återanvänds.
Gör inga förändringar på enheten. Det kan orsaka olyckor eller skada
enheten.
För att mäta blodtrycket måste armen tryckas ihop av manschetten
tillräckligt hårt för att blodflödet genom artären ska stoppas tillfälligt. Det
kan orsaka smärta, domningar eller ett tillfälligt rött märke på armen. Detta
tillstånd uppträder särskilt vid upprepade på varandra följande mätningar.
Eventuella smärtor, domningar eller röda märken försvinner med tiden.
Om blodtrycket mäts för ofta kan det leda till skador på grund av
störningar i blodflödet. Kontrollera att användningen av enheten inte ledertill långvarig försämring av blodcirkulationen när enheten används
upprepade gånger.
Om du genomgått mastektomi ska du kontakta läkare innan du använder
enheten.
Låt inte barn använda enheten själva och använd inte enheten inom
räckhåll för småbarn. Det kan orsaka olyckor eller skador.
Det finns små delar som kan leda till risk för kvävning om små barnsväljer dem av misstag.
Koppla bort nätadaptern när den inte används under mätningen.
Användning av tillbehör som inte beskrivs i bruksanvisningen kanäventyra säkerheten.
Om batteriet kortsluts kan det bli varmt och möjligen orsaka brännskador.
Låt enheten anpassa sig till omgivningen innan den används (ungefär entimme).
Kliniska prövningar har inte genomförts på nyfödda barn och gravida
kvinnor. Får inte användas på nyfödda barn eller gravida kvinnor.
Vidrör inte batterierna, likströmsuttaget och patienten samtidigt. Det kan
leda till elektriska stötar.
Blås inte upp manschetten om den inte sitter runt överarmen.

Kontraindikationer
Följande är försiktighetsåtgärder för korrekt användning av enheten.
Sätt inte manschetten på en arm som är ansluten till annan medicinsk
elektrisk utrustning. Utrustningen fungerar eventuellt inte korrekt.
Personer med allvarlig cirkulationsrubbning i armen måste kontakta en
läkare före användning av enheten för att undvika medicinska problem.
Ställ ingen diagnos själv baserat på mätresultaten och starta ingen
behandling själv. Kontakta alltid läkare för en bedömning av resultaten
och behandlingen.
Sätt inte manschetten på en arm med ett sår som inte läkt.
Sätt inte manschetten på en arm som används för intravenöst dropp eller
blodtransfusion. Det kan orsaka skador eller olyckor.
Använd inte enheten i närheten av lättantändliga gaser som narkosgaser.
Det kan orsaka explosion.
Använd inte enheten i omgivningar med hög koncentration av syrgas, t.ex.
syrgaskammare med högtryck eller syrgastält. Det kan orsaka brand eller
explosion.
Svenska 3Ingående delar
Likströmsuttag
Skärm-knapp-knappSET-knapp

Luftkoppling

START-knapp
Luftanslutning

LuftslangOmfång för korrekt
passform

Positioneringsmärke för artär
ArmmanschettIndexmärke
Batterilucka
Batterifack
1,5 V batterier(R6P, LR6 eller AA)

Skärm

MINNE

Genomsnittliga tryckinställningar

Symbol för IHB/AFib
Symbol för fel manschettstorlek

Systoliskt
tryck

Symbol för rörelsefel
TriCheckTM-symbol

Diastoliskt
tryck

Hjärtsymbol
Indikator för WHO-klassificeringoch tryckindikator

Pulsfrekvens

Batteriindikator
AM-/PM-symbol
Svenska 4

tidsindikator

Symboler
Symboler på enhetens hölje och nätadaptern
Symboler

Funktion/innebörd

Rekommenderad åtgärd

Standby och slå på enheten.
Guide för installation av batteriet
Likström
SN

Serienummer
Tillverkningsdatum
Typ BF: Enheten, manschetten ochslangen är utformade så att de ger
särskilt skydd mot elektriska stötar.
Märkning för medicinteknisk produkt
enligt EU-direktivet
WEEE-märkning
Tillverkare
EU-representant
Likströmsuttagets polaritet

IP

Symbol för internationell skyddsklass
Förvaras torrt

Symboler som visas på skärmen
Symboler

Funktion/innebörd
Rekommenderad åtgärd
Visas när mätning pågår. Blinkar när Mätning pågår. Var så
pulsen identifieras.
stilla som möjligt.
Symbol för IHB/AFib
Visas när hjärtrytmrubbningar
identifieras.
Kan tändas när en mycket liten
vibration som en rysning ellerskakning identifieras.
TriCheckTM-läge
Utför automatiskt tre mätningar i följd
och visar genomsnittliga värden för
de tre mätningarna.
Visas när arm- eller kroppsrörelse
identifieras.
Svenska 5

Mätningen kan ge ett
felaktigt värde. Mät
igen.
Var stilla undermätningen.

Symboler

Funktion/innebörd
Visas under mätningen när
manschetten sitter löst.

Rekommenderad åtgärd
Mätningen kan ge ett
felaktigt värde. Sätt på
manschetten korrektoch mät igen.

Tidigare mätningar sparade i
MINNET.
Genomsnittsvärde
LADDAT BATTERIBatteriströmindikatorn under mätning.
LÅGT BATTERIByt ut alla batterier mot
nya när symbolen
Batteriet är svagt när symbolen
blinkar.
blinkar.
Mät igen.
Ojämnt blodtryck på grund av
Var stilla underrörelser under mätningen.
mätningen.
De systoliska och diastoliska
värdena ligger inom 10 mmHg frånvarandra.
Tryckvärdet ökade inte under
Sätt på manschetten
uppblåsningen.
korrekt och mät igen.
Manschetten sitter inte korrekt.
PUL-FEL
Pulsen identifierades inte korrekt.
Internt fel i blodtrycksmätaren

SYS

Systoliskt blodtryck i mmHg

DIA

Diastoliskt blodtryck i mmHg

PUL

Puls per minut

AM

Data insamlade mellan 4:00 och 9:59

PM

Data insamlade mellan 18:00 och 1:59
Tryckinställningar
Visar tryckvärdet som användaren
har ställt in.
Svenska 6

Ta ut batterierna, tryck på
START-knappen och sättin batterierna igen.
Kontakta återförsäljarenom felet kvarstår.

Driftsläge
1. Normal mätning
Tryck på START-knappen. Blodtrycket mäts och data sparas i enheten.
Enheten kan spara de senaste 90 mätningarna i minnet automatiskt.

2. Hämta data
Tryck på- eller-knappen för att hämta
värdena från minnet. Genomsnitt för alla
värden visas som på bilden till höger.
Varje gång-knappen trycks in, visas
minnesdata enligt följande.
Genomsnittligt
värde
för
alla
morgonmätningar (AM) insamlade mellan
4:00 och 9:59.
Genomsnittligt
värde
för
alla
kvällsmätningar (PM) insamlade mellan
18:00 och 1:59.
Senaste data (i exemplet nr 35)
Sista data (nr 1)
Information om hur man hämtar värden finns i
avsnitt “Hämta minnesdata”. (Se sida 18)

3. Radera alla data som sparats i minnet
Tryck på både- och -knappen i standbyläge.
-symbolen och batteriindikatorn visas. Håll in- och -knappen tills den tända
både-symbolen börjar blinka för att radera alla datasom sparats i minnet.

4. Mäta med önskat systoliskt tryck
På sida 15 visas hur man mäter med ett önskat systoliskt tryck.

Svenska 7

Använda mätaren
Installera/byta batterier
1. Ta bort batteriluckan.
2. Ta ut de förbrukade batterierna ochsätt in nya batterier i batterifacket
som på bilden. Var noga med att
polerna (+ och -) sitter korrekt.
Använd endast batterier av typen
R6P, LR6 eller AA.
3. Montera batteriluckan.

Steg 1
Steg 3

Steg 1
Steg 2

AKTA
Sätt in batterierna i batterifacket som på bilden. Om de sätts in på fel sättfungerar inte enheten.
När(LÅGT BATTERI-indikator) blinkar på skärmen ska alla batterier
bytas ut mot nya. Blanda inte gamla och nya batterier. Det kan förkorta
batteriernas livslängd eller göra så att enheten fungerar på fel sätt.
Vänta i två sekunder eller mer efter att enheten stängts av innan du byter
batterierna.
Utför en blodtrycksmätning om(LÅGT BATTERI-indikator) visas ävenefter att batterierna har bytts. Enheten kan då känna igen de nya batterierna.
(LÅGT BATTERI-indikator) visas inte när batterierna är slut.
Batteriernas livslängd varierar med omgivande temperatur och kan vara
kortare vid låga temperaturer. I allmänhet räcker fyra nya LR6-batterier
ungefär ett år vid mätning två gånger per dag.
Använd endast angivna batterityper. Batterierna som medföljer enheten ärtill för att testa mätarens funktion och kan ha en begränsad livstid.
Ta ut batterierna om enheten inte ska användas under en längre tid.
Batterierna kan läcka och orsaka felfunktion.

Svenska 8

Använda mätaren
Ansluta luftslangen

Luftanslutning

Sätt in luftkopplingen korrekt iluftanslutningen.

Luftkoppling

Ansluta nätadapternSätt i kontakten för nätadaptern ilikströmsuttaget.
Anslut sedan nätadaptern till ett
eluttag.

Kontakt för nätadapter

Likströmsuttag

Använd den specificerade nätadaptern.
(Se sida 23.)Håll i nätadapterns hus och dra utåt när nätadaptern ska kopplas bortfrån eluttaget.
Håll i kontakten för nätadaptern och dra utåt när kontakten för
nätadaptern ska kopplas bort från blodtrycksmätaren.

Svenska 9

Använda mätaren
Ställa in den inbyggda klockan
Ställ in klockan före användning.
1. Tryck på SET-knappen tills året börjar
blinka.
- eller -knappen.
2. Välj år med
Tryck på SET-knappen för att ställa inaktuellt år och gå till månads- och dagsvalet.
Datum kan ställs in mellan 2010 och 2059.

Tryck
eller

År

eller

Månad

- eller -knappen.
3. Välj månad med
Tryck på SET-knappen för att välja aktuellmånad och gå till dagsvalet.

eller

Dag

4. Välj dag med- eller -knappen.
Tryck på SET-knappen för att välja aktuell
dag och gå till tim- och minutvalet.
5. Välj timme med- eller -knappen.
Tryck på SET-knappen för att välja aktuell
timme och gå till minutvalet.
6. Välj minut med- eller -knappen.
(Tryck på SET-knappen för att gå till
tryckinställningsläget. För information, sesida 15.)
Tryck på START-knappen för att stänga av
enheten.

eller

Timme

Minut

Om enheten inte används under tre minuter,stängs den av automatiskt.
Om klockan inte har ställts in,
Stäng avvisar klockan långa streckOmsom på bilden till höger.
START-knappenHåll ner - eller -knappen för att
trycks in, stängsändra värdet kontinuerligt.
enheten av.

eller

Obs!

Svenska 10

tryck
inställningsläge

Använda mätaren
Välja rätt manschettstorlek
Det är viktigt att rätt manschettstorlek används för en korrekt mätning. Om
manschetten inte har rätt storlek kan mätningen ge ett felaktigt
blodtrycksvärde.
Armstorleken är angiven på manschetten.
Indexetoch omfång för korrekt passform på manschetten, visar om duanvänder korrekt manschett. (Se tabellen “Symboler på manschetten” pånästa sida)
Om indexet
pekar utanför omfånget, kontakta den lokala återförsäljaren
för att köpa en ny manschett.
Armmanschetten är en förbrukningsvara. Köp en ny om den blir sliten.
Armstorlek
31 till 45 cm
22 till 42 cm
22 till 32 cm
16 till 24 cm
Armstorlek:

Rekommenderad manschettstorlek
Stor vuxenmanschett
Manschett med stort omfång
Vuxenmanschett
Liten vuxenmanschettOmkretsen vid biceps.

Katalognummer
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Sätta på armmanschetten
1. Lägg manschetten runt överarmen,
ungefär 1–2 cm över armbågens
insida som på bilden.
Placera manschetten direkt mot
huden eftersom klädesplagg kan
göra att pulsen blir svag och leda till
felaktig mätning.

Rulla inte upp ärmen hårt.
Manschett
1–2 cmLuftslang
Positioneringsmärke för artär

2. Åtdragning runt överarmen på
grund av att en skjortärm rullats
upp kan ge felaktiga värden.

Fäst tygfästet
ordentligt.

3. Bekräfta att indexet
pekar inomomfånget för korrekt passform.
Index

Svenska 11

och omfång för korrekt passformAnvända mätaren
Symboler på manschetten
Symboler

Funktion/innebörd

Rekommenderad åtgärd
Placera märket
påöverarmens artär eller i linje
med ringfingret på armens
insida.

Positioneringsmärke för artär

▲
REF

Index
KatalognummerOmfång för korrekt passform
vuxenmanschetten.
Angivet på vuxenmanschetten.

för

AnvänddenstoraÖver omfånget som är angivet på vuxenmanschetten istället
vuxenmanschetten och manschetten för vuxenmanschetten eller
med stort omfång.
manschetten med stortomfång.
Omfång för korrekt passform för
manschetten med stort omfång. Angivet
på manschetten med stort omfång.
Omfång för korrekt passform för den
stora vuxenmanschetten. Angivet påden stora vuxenmanschetten.
Användden
lilla
Under omfånget som är angivet på vuxenmanschetten istället
vuxenmanschetten och manschett med för vuxenmanschetten eller
stort omfång.
manschetten
med
stort
omfång.
Omfång för korrekt passform för liten
vuxenmanschett. Angivet på den lilla
vuxenmanschetten.
Använd vuxenmanschetten
Angivet på den stora vuxenmanschetten istället för den stora
och den lilla vuxenmanschetten.
vuxenmanschetteneller
den lilla vuxenmanschetten.
LotnummerOmfång för korrekt passform
Stor vuxenmanschettA

L

Manschett med stort omfång

S

W

L

Vuxenmanschett

S

A

L

S

A

Liten vuxenmanschettSvenska 12

Använda mätarenSå här görs korrekta mätningar
För noggranna blodtrycksmätningar:
Sitt bekvämt på en stol. Vila armen på bordet. Korsa inte benen. Hållfötterna platt mot golvet och räta på ryggen.
Koppla av i ungefär fem till tio minuter före mätningen.
Placera manschettens mitt i höjd med hjärtat.
Var stilla och tyst under mätningen.
Mät inte omedelbart efter fysisk aktivitet eller bad. Vila i tjugo eller trettio
minuter före mätningen.
Försök att mäta blodtrycket vid samma tidpunkt varje dag.

Mätning
Under mätningen är det normalt att manschetten känns väldigt trång. (Bliinte orolig).

Efter mätningen
Tryck på START-knappen för att stänga av enheten efter mätningen.
Ta bort manschetten och anteckna dina värden. När enheten inte använts
på en minut stängs den av automatiskt.

Svenska 13

Mätningar
Före mätning, läs avsnitt “Anmärkningar för korrekt mätning” på sida 16.

Normal mätning
1.

2.

3.

Placera manschetten på armen (helst vänsterarm).
Var tyst under mätningen.

I höjd med hjärtat
Tryck START

Tryck på START -knappen.
Alla displaysegment visas.
Alla skärmens
Noll (0) blinkar under en kort tid.
segment visas
Skärmen ändras, som visas i figuren till
höger, när mätningen börjar. Manschettenbörjar blåsas upp. Det är normalt att
manschetten känns väldigt trång. En
Visar noll
tryckindikator visas till vänster påStartar uppblåsningskärmen under uppblåsningen.
Obs! Tryck på START -knappen på nytt
om du vill avbryta uppblåsningen.
Trycksätter
När uppblåsningen är klar startas
tömningenautomatisktoch(hjärtsymbolen) blinkar, vilket indikerar att
mätning pågår. När pulsen identifieras
blinkar symbolen vid varje pulsslag.
Obs! Om ett korrekt tryck inte erhålls startar
enheten uppblåsningen igenautomatiskt.
För att undvika ny uppblåsning, seavsnitt “Mäta med SET-trycket” pånästa sida.

4.

5.

Mätning
När mätningen är klar visas det systoliskapågåroch diastoliska trycket samt pulsfrekvensen.
Manschetten släpper ut kvarvarande luft
och töms helt.
Systoliskt tryck
Tryck på START -knappen för att stänga
av enheten.
När enheten inte använts på en minut
stängs den av automatiskt.

Obs! Vänta minst tre minuter mellan
mätningar på samma person.

Svenska 14

Diastoliskt tryck
WHO-klassificering
Pulsfrekvens
Pressar ut kvarvarande
luft automatisktMätningar
UA-1020-W är utformad så att den identifierar pulsen och blåser upp
manschetten till ett systoliskt tryck automatiskt.
Om uppblåsning sker upprepade gånger, gör så här:

Mäta med SET-trycket
Under blodtrycksmätningen kan en ny uppblåsning startas.
Ett fast tryckvärde kan ställas in för att undvika ny uppblåsning.
1.

Tryck på SET-knappen för att öppna
tryckinställningsläget
i
steg
6
av
klockinställningen (se sida 10). De aktuellainställningarna blinkar.

eller

2.

Tryck på - eller -knappen för att välja ett
tryckvärde cirka 30 mmHg eller mer över dittförväntade systoliska tryck bland följande.
AUTO : Automatisk trycksättning (standardvärde)180 : Tryckvärde på 180 mmHg (fast)
210 : Tryckvärde på 210 mmHg (fast)
Stäng av TriCheck
240 : Tryckvärde på 240 mmHg (fast)inställningsläge

3.

Tryck på SET-knappen för att gå till TriCheck-inställningsläget.
Tryck på START-knappen för att stänga av enheten. Om enheten inteanvänds under tre minuter, stängs den av automatiskt.
Nästa mätning utföras med det nya tryckvärdet.

Mäta med önskat systoliskt tryck
Använd den här metoden om uppblåsning sker på nytt upprepade gånger
även om tryckvärdet har ställts in på 240 enligt proceduren ovan eller
när resultaten inte visas även om trycket minskar till 20 mmHg eller mindre.
1.

Placera manschetten på armen (helst vänster arm).

2.

Håll in START -knappen tills ett värdepå cirka 30 till 40 mmHg över dittförväntade systoliska tryck visas.

3.

Släpp START -knappen för att starta
mätningen när önskat värde visas.
Fortsätt att mäta blodtrycket enligt
beskrivningen på föregående sida.

I höjd med
hjärtatHåll in knappen START

Släpp knappen vid
det önskade
systoliska trycket
Se föregående sida
för mätning

Svenska 15

Anmärkningar för korrekt mätning

Sitt i en bekväm position. Lägg armen på ett bord med handflatan uppåt
och manschetten i höjd med hjärtat.

Koppla av i ungefär fem till tio minuter före mätningen. Om du är
upphetsad eller nedstämd av emotionell stress visar mätningen den
stressen som ett högre (eller lägre) blodtrycksvärde än normalt och
pulsvärdet kommer vanligtvis att vara högre än normalt.
En människas blodtryck varierar kontinuerligt beroende på vad man gör
och vad man har ätit. Vad man dricker kan ha en mycket stor och snabb
effekt på blodtrycket.
Enheten baserar mätvärdena på hjärtslagen. Om du har mycket svaga
hjärtslag eller hjärtrytmrubbningar kan enheten att ha svårt att bestämma
ditt blodtryck.
Om enheten identifierar ett onormalt tillstånd avbryts mätningen och en
felsymbol visas. Se sida 6 för en beskrivning av symbolerna.
Blodtrycksmätaren är avsedd att användas av vuxna. Kontakta läkare
innan enheten används på barn. Barn ska inte använda enheten utanövervakning.
Den automatiska blodtrycksmätarens funktion kan påverkas av för hög
temperatur, luftfuktighet eller altitud.

TriCheckTM-mätning
TriCheckTM-läget utför automatiskt tre mätningar i följd och visar
genomsnittsvärden för de tre mätningarna.

Välj TriCheckTM-läget

I tryckinställningsläget:

1.

Tryck
på
SET-knappenitryckinställningsläget för att öppna
TriCheckTM-inställningsläget.
De
aktuella inställningarna blinkar.

2.

Tryck på - eller -knappen för att
slå
på
eller
stängaav
TM
De aktuella inställningarna blinkar
TriCheck -läget.
PÅ: TriCheckTM-läge
eller
AV: Normalt mätningsläge (standardvärde)Ändra inställningen
Tryck på START- eller SET-knappen
för att stänga av enheten. Omeller START
enheten inte används under tre
minuter, stängs den av automatiskt.

3.

Svenska 16

tryck

Mäta i TriCheckTM-läget
1.

2.

3.
4.

Tryck på START -knappen.
Alla displaysegment visas.
Noll (0) visas kortvarigt och den
första mätningen startas.
När mätningen är klar visas det
systoliska och diastoliska trycketsamt
pulsfrekvensen.
Därefterstartas en nedräkning på en minut för
den andra mätningen.
Efter en minut startas den andra
mätningen.

I höjd med hjärtat
START

Tryck
Alla skärmens
segment visas
Första mätning

När mätningen är klar, visas degenomsnittliga mätvärdena för den
första och andra mätningen. Därefter
startas en nedräkning på en minut för
den tredje mätningen.

5.

Efter en minut startas den tredje
mätningen.

6.

När mätningen är klar, visas degenomsnittliga mätvärdena för tre
mätningarna och sparas i minnet.

Obs!
Under
mätningenvisas
TriCheckTM-symbolen.
Tryck på START-knappen för att
avbryta mätningen. I det här falletsparas inga data i minnet.
Avbryt mätningen om symbolen för
fel manschettstorlek visas under den
första mätningen. Fäst manschetten
korrekt och starta en ny mätning.
Efter
mätningensparasgenomsnittsvärdena för de tre
mätningarna i minnet.
Ingen data sparas om mätningarna
avbryts innan den tredje mätningen
har slutförts.

Svenska 17

En minut efter

Resultatet av
första mätningenAndra mätning

En minut efter

Genomsnitt för första
och andra mätningen

Tredje mätning

Genomsnitt för
tre mätningar

Hämta minnesdata
Obs! Enheten sparar automatiskt de senaste nio uppsättningarna i minnet.
1. Tryck på- eller -knappen.
Tryck på eller
Genomsnittet för alla mätningar och
antalet värden visas.
Genomsnittligt systoliskt(Om det inte finns några data, visas “0”.
Genomsnittligt diastoliskt
Tryck på , - eller START-knappen
Genomsnittlig puls
för att stänga av enheten.)
Genomsnitt för
2. Varje gång -knappen (eller
alla data-knappen för att visa värden iomvänd ordningsföljd) trycks in,visas minnesdata enligt följande.
Genomsnittligt
värde
för
alla
morgonmätningar (AM) insamlademellan 4:00 och 9:59.
(10 mätningar i exemplet. Om det
inte finns några data, visas “--”.)
Genomsnittligt värde för alla
kvällsmätningar (PM) insamlademellan 18:00 och 1:59.
(9 mätningar i exemplet. Om det
inte finns några data, visas “--”.)
Senaste data (i exemplet nr 35)
Tre sekunder efter att datanumretvisas, visas mätdata.

Genomsnittligt systoliskt

Genomsnittligt diastoliskt
Genomsnittlig puls

Genomsnitt
för AM-data
Genomsnittligt systoliskt
Genomsnittligt diastoliskt
Genomsnittlig puls

Genomsnitt
för PM-data
Systoliskt
Diastoliskt
Puls

Sista data (nr 1)
Tre sekunder efter att datanumretvisas, visas mätdata.

Senaste data

3. Tryck på -knappen för att gå tillbaka
till genomsnittlig visning av alla
mätvärden efter att sista värdet har
visats.
4. Tryck på START-knappen för att stänga
av enheten. När enheten inte använts
på en minut stängs den av automatiskt.
Svenska 18

Systoliskt
Diastoliskt
Puls

Sista data (äldsta)

Vad är IHB/AFib-indikatorn?
När mätaren identifierar en oregelbunden frekvens under mätningenvisas IHB/AFib-indikatorn på skärmen med mätvärdena.
Obs! Vi rekommenderar att du kontaktar läkare om
IHB/AFib-indikatorn ofta visas.

Vad är AFib (förmaksflimmer)?
Hjärtat drar ihop sig på grund av elektriska signaler i hjärtat och skickar ut
blod genom kroppen. Förmaksflimmer (AFib) inträffar när den elektriska
signalen i förmaket störs och leder till störningar i pulsintervallet.
Förmaksflimmer (AFib) kan leda till att blod stannar i hjärtat och kan lättbilda koagel, vilket är en orsak till stroke och hjärtattack.

Tryckindikator
Indikatorn visar tryckets förlopp under mätningen.
Trycksätter
Släpper ut luft

Uppblåsning pågår

Mätning pågår
Uppblåsning klar

Indikator för WHO-klassificering
Varje segment i indikatorn motsvarar
WHO:sblodtrycksklassificering
enligtbeskrivningen på nästa sida.
Indikator för WHO-klassificering
Allvarlig hypertoniMåttlig hypertoni
Lindrig hypertoni
Hög normal
NormalOptimal: Indikatorn visar ett segment, baserat på
aktuella data, som motsvarar
WHO-klassificeringen.

Svenska 19

Exempel:Måttlig hypertoni

Lindrig hypertoni

Hög normalOm blodtryck
Vad är blodtryck?
Blodtryck är den kraft som hjärtat utövar mot artärväggarna. Systoliskt
tryck är blodtrycket vid hjärtats sammandragning. Diastoliskt tryck är
blodtrycket vid hjärtats utvidgning. Blodtrycket mäts i millimeter
kvicksilver (mmHg). En människas naturliga blodtryck representeras av
det grundläggande trycket som mäts direkt på morgonen när man är
utvilad och innan måltid.

Var är hypertoni och hur kontrolleras det?
Hypertoni – ett onormalt högt artärtryck – kan, om det inte behandlas,orsaka flera hälsoproblem inklusive stroke och hjärtattack. Hypertoni kan
kontrolleras genom att ändra livsstil, undvika stress och med läkemedel
under läkares övervakning.
För att förhindra hypertoni eller hålla det under kontroll:
Rök inte
Motionera regelbundet
Minska intaget av salt och fett
Gör regelbundna hälsokontrollerHåll en lämplig vikt

Varför mäta blodtrycket i hemmet?
När blodtrycket mäts på sjukhus eller läkarmottagning kan det skapaängslan som kan ge ett förhöjt värde, 25 till 30 mmHg högre än det som
mäts i hemmet. Mätning i hemmet minskar effekterna av yttre påverkanpå blodtrycksvärdena, utgör ett komplement till läkarens mätningar och
ger en mer exakt och fullständig blodtryckshistorik.

WHO:s blodtrycksklassificering

Variationer i blodtrycket

Diastoliskt blodtryck

Standarder för bedömning av Referensmaterial: Journal of Hypertension 1999, Vol 17 No. 2
högt blodtryck, oavsett ålder, mmHg
Hypertoni grad 3 (allvarlig)
110
har
upprättats
av
Hypertoni grad 2 (måttlig)
105
Världshälsoorganisationen
100(WHO),
somvisasi
Hypertoni grad 1 (lindrig)
95
diagrammet.
90
85
80

Hög-normal

NormalOptimal120 130 140 150 160 170 180
Systoliskt blodtryck
mmHg

En människas blodtryck
varierar mycket dag till dag
och beroende på årstid. Det
kan variera med 30 till 50 mmHg på grund av olika förutsättningar underdagen. Hos människor med hypertoni är variationerna ännu större.
Normalt stiger blodtrycket vid arbete eller lek och sjunker till sin lägsta
nivå när man sover. Bli därför inte alltför oroad över resultaten från en
mätning.
Svenska 20

Typiska fluktuationer under en dag(mätt var femte minut)
Systoliskt

Blodtryck

Mät vid samma tid varje dag med metoden
som beskrivs i bruksanvisningen så att du
lär
känna
ditt
normalablodtryck.
Regelbundna mätningar ger en mer
fullständig blodtryckshistorik. Var noga med
att notera datum och tid när du mäter dittblodtryck. Kontakta läkare för att tolka dina
blodtrycksvärden.

Sömn

Diastoliskt

PM

AM

Tid

Felsökning
Problem
Ingenting visas
på skärmen,även när
enheten ärpåslagen.

Möjlig orsak
Batterierna är slut.
Batteriernas poler ärinte i rätt position.

Batterispänningen är för
låg.
(LÅGT
Manschetten
BATTERI-indikator)blåses inte upp.
blinkar. Om batterierna
är helt urladdade visas
inte symbolen.
Manschetten sitter intekorrekt.
Du rörde armen eller
kroppen undermätningen.
Enheten mäter
inte.
Manschettens position
Värdena är för
är inte korrekt.
höga eller för
låga.

Rekommenderad åtgärd
Byt ut alla batterierna mot nya.
Sätt in batterierna så att den
negativa och den positiva polen
matchar de som visas ibatterifacket.

Byt ut alla batterierna mot nya.

Sätt på manschetten korrekt.
Var stilla och tyst undermätningen.
Sitt bekvämt och stilla.
Lägg armen på ett bord med
handflatan uppåt och
manschetten i höjd med hjärtat.
Om du har mycket svaga
hjärtslag eller
hjärtrytmrubbningar kan
enheten att ha svårt att
bestämma ditt blodtryck.

Värdet skiljer sig fråndet som uppmätts på en Se avsnitt “Varför mäta
blodtrycket i hemmet?”.
klinik ellerÖvrigtläkarmottagning.
Ta ut batterierna. Sätt tillbaka
dem korrekt och mät igen.
Obs! Kontakta återförsäljaren om åtgärderna som beskrivs ovan inte löser
problemet. Försök inte att öppna eller reparera produkten – garantin
gäller då inte längre.
Svenska 21

Underhåll
Öppna inte enheten. Den har ömtåliga elektriska komponenter och en
invecklad luftenhet som kan skadas. Om problemet inte kan lösas med
felsökningsanvisningarna, kontakta den auktoriserade återförsäljaren i dittområde eller vår kundservice. Kundservice hos A&D ger teknisk information,
tillhandahåller reservdelar och enheter till auktoriserade återförsäljare.
Enheten har utformats och tillverkats för att få en lång livslängd. Vi
rekommenderar emellertid normalt att låta inspektera enheten vartannat
år för att säkerställa korrekt funktion och noggrannhet. Kontakta den
auktoriserade återförsäljaren i ditt område eller A&D för underhåll.

Tekniska data
Typ

UA-1020-W

Mätmetod

Oscillometrisk mätning

Mätintervall

Tryck:
0–299 mmHg
Systoliskt tryck: 60–279 mmHg
Diastoliskt tryck: 40–200 mmHg
Puls:
40–180 slag/minut

Mätnoggrannhet

Tryck:
Puls:

Strömförsörjning

4 x V 1,5 batterier (R6P, LR6 eller AA) eller
Nätadapter (tillval) (TB-233C) (medföljer inte)

Antal mätningar

Ca 1 000 gånger
LR6 (alkaliska batterier)Cirka 300 gånger
R6P (manganbatterier)
Med ett tryckvärde på 180 mmHg vid
rumstemperatur (23 °C)

Klassificering

Internt strömförsörjd medicinteknisk utrustning(drivs med batterier)/
Klass II (drivs med nätadapter)
Kontinuerligt driftläge

Kliniskt test

I enlighet med ISO 81060-2: 2013
I den kliniska valideringsstudien användes K5 på
85 subjekt för bestämning av diastoliskt blodtryck

EMD

IEC 60601-1-2: 2014

Minne

Senaste 90 mätningarna

Driftförhållanden
Transport-/förvaringsförhållandenMått+10 till +40 °C/15 till 85 % RH/800 till 1 060 hPa-20 till +60 °C/10 till 95 % RH/700 till 1 060 hPa±3 mmHg±5 %Ca 140 [B] x 60 [H] x 105 [D] mm
Svenska 22

Vikt

Ca 285 g, exklusive batterier

Applicerad delManschett Typ BF

Kapslingsklass

Enhet: IP20

Användbar livstid

Enhet: 5 år (vid användning sex gånger per dag)
Manschett: 2 år (vid användning sex gånger per
dag)
Nätadapter: 5 år (vid användning sex gånger per
dag)

Nätadapter som tillbehör

Adaptern är avsedd för att ansluta
blodtrycksmätare till en strömkälla i hemmet.

TB-233C

Kontakta lokala A&D-återförsäljaren för inköp.
Nätadaptern måste inspekteras eller bytas ut
regelbundet.
Symboler på nätadaptern
Symboler
Funktion/innebörd
Endast för inomhusbruk
Enhet av klass II
Termisk säkring
Säkring
Märkning för medicinteknisk produkt
enligt EU-direktivet
Märkning för certifiering enligt EAC
Polaritet för kontakten för nätadaptern

Tillbehör säljs separat
Manschett
Katalognummer
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Manschettstorlek
Stor vuxenmanschett
Manschett med stortomfångVuxenmanschett
Liten vuxenmanschettArmstorlek
31 till 45 cm
22 till 42 cm
22 till 32 cm
16 till 24 cm

Nätadapter
Katalognummer
TB-233C

Obs!

Kontakt
Typ CSpecifikationer kan ändras utan föregående meddelande.
IP-klassificeringen är den nivå av skydd som kapslingen ger enligt IEC
60529. Enheten är skyddad mot fasta främmande föremål med 12 mm
diameter och större, t.ex. ett finger. Enheten är inte skyddad mot vatten.

Svenska 23

ANTECKNINGAR

Svenska 24

Sisällys
Hyvät asiakkaat ..................................................................................... 2
Ennakkohuomautukset .......................................................................... 2
Varotoimet ............................................................................................ 2
Kontraindikaatiot ............................................................................. 3
Osien tunnistus ..................................................................................... 4
Symbolit ................................................................................................ 5
Toimintatila........................................................................................... 7
Monitorin käyttö.................................................................................... 8
Paristojen asennus/vaihto ............................................................... 8
Ilmaletkun liittäminen ...................................................................... 9
Vaihtovirtasovittimen liittäminen ..................................................... 9
Sisäänrakennetun kellon säätäminen ............................................. 10
Oikean mansettikoon valitseminen ................................................ 11
Käsivarren mansetin asettaminen .................................................. 11
Tarkkojen mittausten tekeminen ................................................... 13
Mittaus .......................................................................................... 13
Mittauksen jälkeen ........................................................................ 13
Mittaukset ........................................................................................... 14
Tavallinen mittaus ......................................................................... 14
Mittaus ASETETULLA paineella ....................................................... 15
Mittaus halutulla systolisella paineella ........................................... 15
Tarkkaa mittausta koskevia huomautuksia .................................... 16
TriCheckTM-mittaus ............................................................................. 16
Tietojen hakeminen muistista ............................................................. 18
Mikä on IHB/AFib-indikaattori? ........................................................... 19
Mikä on AFib? ...................................................................................... 19
Paineen palkki-indikaattori ................................................................. 19
WHO-luokitusindikaattori.................................................................... 19
Tietoja verenpaineesta ........................................................................ 20
Mikä on verenpaine? ...................................................................... 20
Mitä kohonnut verenpaine tarkoittaa ja kuinka sitä hoidetaan? ...... 20
Miksi verenpainetta kannattaa mitata kotona? ............................... 20
WHO:n verenpaineluokitus............................................................. 20
Verenpaineen vaihtelut .................................................................. 20
Vianmääritys ....................................................................................... 21
Huolto ................................................................................................. 22
Tekniset tiedot .................................................................................... 22

Suomi 1Hyvät asiakkaat
Onnittelut! Olette hankkineet A&D:n huipputekniikkaa edustavan
verenpainemonitorin. Tämä helppokäyttöinen ja tarkka monitori helpottaa
päivittäistä verenpainehoitoa.
Suosittelemme lukemaan tämän oppaan huolellisesti ennen laitteen
käyttöä ensimmäistä kertaa.

Ennakkohuomautukset
Tämä laite on lääkinnällisiä laitteita koskevan neuvoston direktiivin 93/42-yhdenmukaisuusmerkintä.
ETY mukainen. Tämän osoittaa(0123: Numero viittaa asianomaiseen ilmoitettuun laitokseen)
Laite on tarkoitettu aikuisille, ei vastasyntyneille tai pikkulapsille tehtyihin
mittauksiin.
Käyttöympäristö. Laite on tarkoitettu ostajan omaan käyttöön
kotihoitoympäristöön.
Tämä laite on tarkoitettu ihmisten verenpaineen ja sydämen
lyöntitiheyden diagnostiseen mittaamiseen.

Varotoimet
Tämän laitteen valmistuksessa on käytetty tarkkuuskomponentteja.
Laitteen altistuminen äärimmäisille lämpötiloille, kosteudelle, suoralleauringonvalolle, iskuille tai pölylle pitää estää.
Puhdista laite ja mansetti kuivalla, pehmeällä liinalla tai vedellä
kostutetulla liinalla ja neutraalilla pesuaineella. Älä koskaan käytä
alkoholia, bentseeniä, ohennusainetta tai muita voimakkaita kemikaaleja
laitteen tai mansetin puhdistamiseen.
Vältä mansetin taittamista tiukalle tai letkun säilyttämistä pitkiä aikoja
tiukkaan käärittynä, jottei osien käyttöikä lyhene.
Varmista, etteivät vauvat tai pikkulapset voi kuristua letkuun ja johtoon.
Älä kierrä ilmaletkua mittauksen aikana. Se voi aiheuttaa
loukkaantumisen mansetin jatkuvan paineen vuoksi.
Laite ja mansetti eivät ole vedenkestäviä. Älä anna sateen, hien tai veden
kastella laitetta tai mansettia.
Mittaus voi vääristyä, jos laitetta käytetään lähellä televisiota,
mikroaaltouunia, matkapuhelinta, röntgensäteitä tai muita laitteita, joillaon voimakas sähkökenttä.
Langattomat viestintälaitteet, kuten kotiverkkolaitteet, matkapuhelimet,
langattomat puhelimet ja niiden tukiasemat sekä radiopuhelimet voivat
vaikuttaa verenpainemonitoriin.
Tämän vuoksi näihin laitteisiin on pidettävä vähintään 30 cm etäisyys.
Käytettyjä laitteita, osia ja paristoja ei saa käsitellä tavallisena
kotitalousjätteenä. Ne on hävitettävä voimassa olevien paikallisten
säädösten mukaan.
Kun käytössä on vaihtovirtasovitin, varmista, että vaihtovirtasovittimen voi
Suomi 2

tarvittaessa irrottaa nopeasti pistorasiasta.
Kun käytät laitetta uudelleen, varmista, että se on puhdas.
Älä tee laitteeseen muutoksia. Se voi johtaa onnettomuuksiin tai laitteen
vaurioitumiseen.
Verenpaineen mittaamiseksi mansetin pitää puristaa käsivartta tarpeeksi
voimakkaasti niin, että veren virtaus valtimoon keskeytyy väliaikaisesti.
Tämä voi aiheuttaa kipua, tunnottomuutta tai väliaikaisen punaisen jäljen
käsivarteen. Näin tapahtuu etenkin silloin, jos mittaus tehdään useamman
kerran peräkkäin.
Kipu, tunnottomuus ja punaiset jäljet häviävät ajan myötä.
Verenpaineen mittaaminen liian usein voi olla haitallista veren virtauksen
häiriintymisen vuoksi. Kun laitetta käytetään toistuvasti, tarkista, ettei sen
käyttö johda veren virtauksen pitkäaikaiseen heikentymiseen.
Jos sinulle on tehty mastektomia, kysy lääkäriltä ohjeita ennen laitteen
käyttöä.
Älä anna lasten käyttää laitetta ilman valvontaa, äläkä käytä laitetta
paikassa, jossa se on pikkulasten ulottuvilla. Se voi johtaa
onnettomuuksiin tai vaurioihin.
Laite sisältää pieniä osia. Ne voivat aiheuttaa tukehtumisvaaran, mikäli
pikkulapset vahingossa nielaisevat ne.
Kytke vaihtovirtasovitin irti, jos sitä ei käytetä mittauksen aikana.
Muiden kuin tässä oppaassa mainittujen lisävarusteiden käyttö voi olla
turvallisuusriski.
Jos paristo joutuu oikosulkuun, se voi kuumentua ja aiheuttaa palovammoja.
Anna laitteen sopeutua ympäristöön ennen sen käyttöä (noin tunnin ajan).
Kliinisiä testauksia ei ole tehty vastasyntyneiden lasten tai raskaana
olevien naisten osalta. Älä käytä laitetta vastasyntyneiden lasten tai
raskaana olevien naisten hoitoon.
Älä koske paristoihin, DC-liittimeen ja potilaaseen samanaikaisesti. Se voi
aiheuttaa sähköiskun.
Älä täytä mansettia, jos sitä ei ole kääritty olkavarren ympärille.

Kontraindikaatiot
Seuraavaksi kuvataan varotoimet laitteen oikeaa käyttöä varten.
Älä laita mansettia käsivarteen, jossa on jo toinen sähköinen
lääkinnällinen laite. Laite ei silloin ehkä toimi kunnolla.
Terveysongelmien välttämiseksi henkilöiden, joilla on käsivarren vakava
verenkiertohäiriö, tulee kysyä lääkäriltä ohjeita ennen laitteen käyttöä.
Älä tee itse diagnoosia mittaustulosten perusteella äläkä aloita hoitoa
omatoimisesti. Pyydä aina lääkäriä arvioimaan tulokset ja hoidontarve.
Älä laita mansettia käsivarteen, jossa on parantumaton haava.
Älä laita mansettia käsivarteen, johon tehdään suonensisäistä tiputusta
tai verensiirtoa. Se voi johtaa vammoihin tai onnettomuuksiin.
Älä käytä laitetta tilassa, jossa on syttyviä kaasuja, kuten
anestesiakaasuja. Se voi aiheuttaa räjähdyksen.
Älä käytä laitetta ympäristöissä, joissa on suuria happipitoisuuksia, kuten
korkeapaineisessa happikammiossa tai happiteltassa. Se voi johtaa
tulipaloon tai räjähdykseen.
Suomi 3

Osien tunnistus
DC-liitin

Näyttö

-painike-painike
SET-painike

Ilmaliitospistoke

START-painike
Ilmaliitäntä
Ilmaletku
Oikea sopivuusalue

Valtimon paikkamerkki
Käsivarren
mansetti

Osoitinmerkki

Paristolokeron
kansi

Paristolokero

1,5 V:n paristot(R6P, LR6 tai AA)

NäyttöMUISTI

Keskiarvo

Paineasetukset

IHB/AFib-symboli
Mansetin sopivuusvirheen symboli

Systolinen
paine

Liikkeestä johtuvan virheen symboli

TriCheckTM-symboli

Diastolinen
paine

Sydänmerkki
WHO-luokitusindikaattorija paineen palkki-indikaattori
Pariston indikaattori

Sydämenlyöntitiheys

AM / PM -merkki
Suomi 4

Kellonäyttö

Symbolit
Laitekoteloon ja vaihtovirtasovittimeen merkityt symbolit
Symbolit

Toiminta/Tarkoitus
Valmiustila ja laitteen päälle kytkentä.

Suositeltu toimi

Pariston asennusohje
Tasavirta
SN

Sarjanumero
Valmistuspäivä
Tyyppi BF: Laitteessa, mansetissa
ja letkuissa on suojaus sähköiskuja
vastaan.
EY:n direktiivin mukaisen
lääkinnällisen laitteen merkintä
WEEE-merkintä
Valmistaja
Edustaja EU:ssa
DC-liittimen napaisuus

IP

Kansainvälinen suojausmerkintä
Pidä kuivana

Näytössä näkyvät symbolit
Symbolit

Toiminta/Tarkoitus
Suositeltu toimi
Tulee näkyviin mittauksen aikana. Mittaus on käynnissä. Liiku
Vilkkuu, kun pulssi löytyy.
mahdollisimman vähän.
IHB/AFib-symboli
Tulee näkyviin, kun havaitaan
epäsäännöllinen sydämensyke.
Se saattaa syttyä, jos havaitaan
hienoista tärinää, kuten vapinaa tai
puistatusta.
TriCheckTM-tila
Ottaa automaattisesti kolme
peräkkäistä mittausta ja näyttää
näiden kolmen mittauksen
keskiarvon.
Tulee näkyviin, kun havaitaan
vartalon tai käsivarren liike.
Suomi 5

Mittaustuloksen arvo voi
olla virheellinen. Mittaa
uudelleen.
Pysy mittauksen aikana
liikkumatta.

Symbolit

Toiminta/Tarkoitus
Tulee näkyviin mittauksen aikana,jos mansetti on asetettu löysästi.

Suositeltu toimi
Mittaustuloksen arvo voi
olla virheellinen. Kiinnitä
mansetti oikein ja mittaa
uudelleen.

Aiemmat muistiin (MEMORY)
tallennetut mittaukset.
Keskiarvotiedot
PARISTO TÄYSI
Pariston virtaindikaattori
mittauksen aikana.
PARISTO VÄHISSÄ
Vilkkuu, kun pariston virta on
vähissä.
Epävakaa verenpaine mittauksen
aikaisen liikkumisen vuoksi.

Vaihda kaikki paristot
uusiin, kun merkki vilkkuu.
Yritä mitata uudelleen.
Pysy mittauksen aikana
liikkumatta.

Systolisen ja diastolisen arvon
välinen ero on enintään 10 mmHg.
Painearvo ei noussut täytön aikana.
Mansettia ei ole asetettu oikein.

Kiinnitä mansetti oikein ja
yritä mitata uudelleen.

PUL DISPLAY ERROR
Pulssia ei havaita oikein.

DIA

Poista paristot, paina
START -painiketta ja
asenna sitten paristot
Verenpainemonitorin sisäinen virhe
uudelleen. Jos virhe
näytetään yhä, ota yhteys
jälleenmyyjään.
Systolinen verenpaine yksikössä
mmHg
Diastolinen verenpaine yksikössä
mmHg

PUL

Sykettä minuutissa

AM

Tiedot kerätty klo 4:00–9:59

PM

Tiedot kerätty klo 18:00–1:59

SYS

Paineasetukset
Ilmaisee käyttäjän aiemmin
asettaman painearvon.
Suomi 6

Toimintatila
1. Tavallinen mittaus
Paina START -painiketta. Verenpaine mitataan ja tiedot tallennetaan
laitteelle. Laite voi tallentaa muistiin 90 tietosarjaa automaattisesti.

2. Tietojen hakeminen
Hae muistissa olevat tiedot painamallatai-painiketta. Näkyville ilmestyy kaikkien
mittausten keskiarvo oikeanpuoleisessa
kuvassa osoitetulla tavalla.
Tämä jälkeen jokaisella-painikkeen
painalluksella laitteen näyttöön tulevatseuraavat.
Kaikkien aamulla (AM) klo 4:00–9:59
suoritettujen mittausten keskiarvo.
Kaikkien illalla (PM) klo 18:00–1:59
suoritettujen mittausten keskiarvo.
Viimeisimmät tiedot (nro n, esimerkissä
nro 35)
Viimeisimmät tiedot (nro 1)
Tarkemmat
tiedottietojen
muistista
hakemisesta löytyvät osiosta “Tietojen
hakeminen muistista”. (Katso sivu 18.)

3. Kaikkien muistissa olevien tietojen poistaminen
Paina sekä
- että -painiketta. Näkyvilleilmestyy-merkki ja pariston indikaattori.
- ja -painikkeita, kunnes
Paina pitkään
syttynyt-merkki alkaa vilkkua kaikkien
muistissa olevien tietojen poistamiseksi.

4. Mittaus halutulla systolisella paineella
Katso sivulta 15 tiedot mittaamisesta halutulla systolisella paineella.
Suomi 7

Monitorin käyttö
Paristojen asennus/vaihto
1. Irrota paristolokeron kansi.
2. Poista käytetyt paristot ja asenna
uudet paristot paristolokeroon kuvan
mukaisesti.
Varmista,että
napaisuudet (+ ja -) asetetaan oikein.
Käytä vain paristoja R6P, LR6 taiAA.
3. Asenna paristolokeron kansi.

Vaihe 1
Vaihe 3Vaihe 1

Vaihe 2

HUOMAUTUS
Laita paristot paristolokeroon kuvatulla tavalla. Jos ne laitetaan väärin, laite
ei toimi.
Kun(PARISTO VÄHISSÄ -merkki) vilkkuu näytössä, vaihda kaikki
paristot uusiin. Älä sekoita vanhoja ja uusia paristoja. Se voi lyhentää
paristojen käyttöikää tai aiheuttaa laitteeseen toimintahäiriön.
Vaihda paristot, kun laitteen sammumisesta on kulunut kaksi sekuntia tai
enemmän.
Jos(LOW BATTERY -merkki) näkyy, vaikka paristot on vaihdettu, teeverenpainemittaus. Laite voi tällöin tunnistaa uudet paristot.
(PARISTO VÄHISSÄ -merkki) ei tule näkyviin, kun paristot ovat tyhjät.
Paristojen käyttöikä vaihtelee ympäristön lämpötilan mukaan. Se voi ollalyhempi alhaisissa lämpötiloissa. Yleensä neljä uutta LR6-paristoa kestävät
noin yhden vuoden, kun niitä käytetään mittaukseen kahdesti päivässä.
Käytä vain määritettyjä paristoja. Laitteen mukana toimitetut paristot ontarkoitettu monitorin toiminnan tarkastamiseen, joten niiden käyttöikä voi
olla lyhyempi.
Poista paristot, jos laitetta ei tulla käyttämään pitkään aikaan.
Paristot saattavat vuotaa ja aiheuttaa laitteeseen toimintahäiriön.

Suomi 8

Monitorin käyttö
Ilmaletkun liittäminen

Ilmaliitäntä

Asenna ilmaliitospistoke tiukasti
ilmaliitäntään.

Ilmaliitospistoke

Vaihtovirtasovittimen liittäminen
Kytke vaihtovirtasovittimen pistoke
DC-liittimeen.
Liitä
sittenvaihtovirtasovitin
pistorasiaan.

Vaihtovirtasovittimen
pistokeDC-liitin
Käytä määritettyä vaihtovirtasovitinta.
(Katso sivu 23.)
Kun irrotat vaihtovirtasovittimen pistorasiasta, tartu vaihtovirtasovittimen
runkoon ja vedä se ulos pistorasiasta.
Kun
irrotatvaihtovirtasovittimen
verenpainemonitorista,tartu
vaihtovirtasovittimen pistokkeeseen ja vedä se ulos monitorista.

Suomi 9

Monitorin käyttö
Sisäänrakennetun kellon säätäminen
Säädä kello ennen käyttöä.
1. Paina SET -painiketta, kunnes vuosi alkaa
vilkkua.
- tai -painikkeella.
2. Valitse vuosi
Aseta nykyinen vuosi ja siirry valitsemaan
kuukausi/päivä painamalla SET -painiketta.
Päivämäärä
voidaanasettaa
välillä
2010–2059.
- tai -painikkeella.
3. Valitse kuukausi
Aseta nykyinen kuukausi ja siirry valitsemaan
päivä painamalla SET -painiketta.
- tai -painikkeella.
4. Valitse päivä
Aseta
nykyinenpäivä
jatunti/minuutti-valintaan painamalla
SET -painiketta.

Paina
Vuosi

tai

tai
Kuukausi

Päivä

tai

siirrytai
Tunti

5. Valitse tunti- tai -painikkeella.
Aseta nykyinen tunti ja siirry valitsemaanminuutti painamalla SET -painiketta.
6. Valitse minuutti- tai -painikkeella.
(Siirry paineen asetustilaan painamalla
SET -painiketta. Katso tiedot sivulta 15.)
Sammuta laite painamalla START -painiketta.

MinuuttitaiHuom.: Laite sammuu automaattisesti, jos sitä ei
käytetä kolmen minuutin kuluessa.
Sammutus
Kun kelloa ei ole asetettu,
kellonäytöllä näkyy viivat
Laitesammuu paine
oikealla kuvatulla tavalla.
millointahansa asetustila
Kun painat- tai-painiketta
painamalla
START-painiketta.
pitkään, arvo muuttuu jatkuvasti.

Suomi 10

Monitorin käyttö
Oikean mansettikoon valitseminen
On tärkeää käyttää oikeankokoista mansettia, jotta mittaustulokset olisivat
tarkkoja. Jos mansetti ei ole oikeankokoinen, verenpainemittauksen tulos
voi olla virheellinen.
Käsivarren koko on merkitty jokaiseen mansettiin.
Mansettiin merkitty osoitin
ja oikea sopivuusalue kertovat, onko
käytössä oikea mansetti. (Katso seuraavalla sivulla oleva taulukko“Mansettiin merkityt symbolit”.)
Jos osoitin
osuu alueen ulkopuolelle, ota yhteyttä paikalliseen
jälleenmyyjään ja hanki toinen mansetti.
Käsivarren mansetti on kuluva tuote. Jos se on kulunut, hanki tilalle uusi.
Käsivarren koko
Suositeltu mansetin koko
31 cm–45 cm
Suuri aikuisten mansetti
22 cm–42 cm
Laajan kokoalueen mansetti
22 cm–32 cm
Aikuisten mansetti
16–24 cm
Pieni aikuisten mansetti
Käsivarren koko: Ympärysmitta hauiksen kohdalla.

Kataloginumero
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Käsivarren mansetin asettaminen
1. Kääri mansetti olkavarren ympärille,
noin 1–2 cm kyynärpään sisäpuolen
yläpuolelle, kuten kuvassa.
Laita mansetti suoraan iholle.
Vaatteet voivat heikentää pulssin
tuntumista ja aiheuttaa siksi
mittausvirheen.

Älä rullaa paidanhihaa kireälle.
Mansetti
1–2 cmIlmaletku
Valtimon paikkamerkki

2. Jos tiukalle kääritty hiha puristaa
olkavartta, tarkkoja mittaustuloksia
ei ehkä saada.

Kiinnitäkangaskiinnike
tiukasti.

3. Varmista, että osoitin
osuu
oikean sopivuusalueen sisään.
Osoitin

Suomi 11

ja oikea sopivuusalue

Monitorin käyttö
Mansettiin merkityt symbolit
Symbolit
Toiminta/Tarkoitus

Suositeltu toimi
Aseta-merkki olkavarren
valtimon päälle tai käsivarren
sisäpuolella samaan linjaan
nimettömän kanssa.

Valtimon paikkamerkki▲
REF

Osoitin
Kataloginumero
Aikuistenmansetinoikea
sopivuusalue.
Merkitty aikuisten mansettiin.
suurtaaikuisten
Alueen yli, merkitty aikuisten Käytä
mansettia
aikuistenmansetinmansettiin ja laajan kokoalueen tailaajankokoalueen
mansettiin.
mansetin sijaan.
Laajan kokoalueen mansetin oikea
sopivuusalue.
Merkittylaajankokoalueen mansettiin.
Suuren aikuisten mansetin oikea
sopivuusalue. Merkitty suureen
aikuisten mansettiin.
pientä
aikuisten
Alueen alle, merkitty aikuisten Käytä
mansettia
aikuistenmansetinmansettiin ja laajan kokoalueen tailaajankokoalueen
mansettiin.
mansetin sijaan.
Pienen aikuisten mansetin oikea
sopivuusalue. Merkitty pieneen
aikuisten mansettiin.
Merkitty suureen ja
aikuisten mansettiin.

aikuisten mansettia
pieneen Käytä
suuren tai pienen aikuistenmansetin sijaan.

Eränumero
Oikea sopivuusalue
Suuri aikuisten mansetti

A

L

Laajan kokoalueen
mansetti

S

W

L

Aikuisten mansetti

S

A

L

S

A

Pieni aikuisten mansetti
Suomi 12

Monitorin käyttö
Tarkkojen mittausten tekeminen
Verenpainemittaus antaa tarkimmat tulokset, kun toimitaan seuraavasti:
Istu mukavasti tuoliin. Laita käsivarsi lepäämään pöydälle. Älä laita jalkoja
ristiin. Pidä jalat tasaisesti lattiassa ja suorista selkä.
Rentoudu viidestä kymmeneen minuuttia ennen mittauksen tekemistä.
Laita mansetin keskikohta sydämen tasolle.
Pysy paikallasi ja puhumatta mittauksen ajan.
Älä mittaa välittömästi fyysisen rasituksen tai kylvyn jälkeen. Lepää
kaksikymmentä tai kolmekymmentä minuuttia ennen mittauksen tekemistä.
Pyri mittaamaan verenpaine joka päivä samaan kellonaikaan.

Mittaus
On normaalia, että mansetti tuntuu mittauksen aikana hyvin tiukalta. (Äläsäikähdä.)

Mittauksen jälkeen
Sammuta laite mittauksen jälkeen painamalla START -painiketta.
Poista mansetti ja kirjaa tiedot muistiin. Laite sammuu automaattisesti, jossitä ei käytetä minuutin kuluessa.

Suomi 13

Mittaukset
Lue ennen mittausta sivun 16 osio “Tarkkaa mittausta koskevia
huomautuksia”.

Tavallinen mittaus
1.

Laita mansetti käsivarteen
vasempaan).
Istu hiljaa mittauksen ajan.

2.

Paina
Paina START -painiketta.
Kaikki näytön alueet ovat näkyvissä.
Nolla (0) näkyy hetken vilkkuvana.
Näyttö muuttuu mittauksen alkaessa
Kaikki näytön
oikeanpuoleisessa kuvassa osoitetulla
tavalla. Mansetti alkaa täyttyä. On alueet ovat näkyvissä
normaalia, että mansetti tuntuu hyvintiukalta.
Täytönaikananäytönvasemmassa reunassa näkyy paineen
palkki-indikaattori.
Nolla-näyttö
Huom.: Täytön voi keskeyttää milloin
tahansa painamalla uudelleen
Täytön aloitus
START -painiketta.

3.

Kun mansetti on täytetty, se alkaa
tyhjentyä
automaattisestija(sydänmerkki) vilkkuu ilmoittaen, että
mittaus on käynnissä. Kun pulssi löytyy,
merkki vilkkuu jokaisella sykkeellä.
Huom.: Jos riittävää painetta ei saavuteta,mansetti alkaa täyttyä uudelleen
automaattisesti.
Katso uudelleen täyttämisen
estäminen seuraavan sivun
osiosta “Mittaus ASETETULLA
paineella”.

4.

5.

(mieluiten

Kun mittaus on valmis, systolisen ja
diastolisen paineen lukemat sekä sydämen
lyöntitiheys tulevat näyttöön.
Mansettiin jäänyt ilma poistetaan ja setyhjenee täysin.
Sammuta laite painamalla START-painiketta.
Laite sammuu automaattisesti, jos sitä
ei käytetä minuutin kuluessa.

Huom.: Odota vähintään kolme minuuttia
ennen mittauksen tekemistä samalle
henkilölle.
Suomi 14

Sydämen tasolla

START

Paineistus

Mittauskäynnissä
Systolinen paine
Diastolinen paine
WHO-luokitus
Sydämen lyöntitiheys
Poistaa jäljelle jääneen
ilman automaattisestiMittaukset
UA-1020-W on suunniteltu etsimään pulssi ja täyttämään mansetti
systolisen paineen tasolle automaattisesti.
Jos mansetti täytetään toistuvasti, käytä seuraavia menetelmiä.

Mittaus ASETETULLA paineella

Mansetti saattaa täyttyä uudelleen verenpainemittauksen aikana.
Voit välttää uudelleen täytön asettamalla kiinteän painearvon.

1.

Siirry paineen asetustilaan painamalla kellon
asetustoimenpiteen vaiheessa 6 (katso sivu
10) SET -painiketta. Nykyinen asetus vilkkuu.

2.

Valitse painearvoksi odotettua systolista
painettasi
noin
30 mmHgsuurempi- taipainearvo tai enemmän painamalla-painiketta seuraavista.

3.

tai

AUTO : Automaattinen paineistus (oletusarvo)
180 : Painearvo 180 mmHg (kiinteä)
210 : Painearvo 210 mmHg (kiinteä)
240 : Painearvo 240 mmHg (kiinteä)
Sammutus TriCheckasetustila
Siirry TriCheck-asetustilaan painamalla SET-painiketta.
Sammutalaite
painamalla
START-painiketta.
Laitesammuuautomaattisesti, jos sitä ei käytetä kolmen minuutin kuluessa.
Seuraava mittaus suoritetaan uudella painearvolla.

Mittaus halutulla systolisella paineella

Käytä tätä menetelmää, jos mansetti täyttyy toistuvasti uudelleen, vaikka
painearvoksi on asetettu edellä kuvatulla toimenpiteellä 240 tai jos
tuloksia ei näytetä, vaikka paine laskee arvoon 20 mmHG taialhaisemmaksi.

1.

Laita mansetti käsivarteen (mieluiten vasempaan). Sydämen tasolla

2.

Paina
pitkään
START -painiketta,
kunnes näkyviin tulee lukema, joka on
noin 30–40 mmHg korkeampi kuin
odotettu systolinen paine.

3.

Kun haluttu numero saavutetaan,
aloitamittausvapauttamalla
START -painike. Jatka verenpaineen
mittausta edellisellä sivulla kuvatulla
tavalla.

Paina ja pidä
painettuna painike START
Vapauta painike
halutun systolisen
paineen kohdalla

Katso mittauksen tiedot
edelliseltä sivulta
Suomi 15

Tarkkaa mittausta koskevia huomautuksia

Istu alas mukavaan asentoon. Laita käsivarsi pöydälle niin, että kämmen
on ylöspäin ja mansetti on sydämen tasolla.
Rentoudu viidestä kymmeneen minuuttia ennen mittauksen tekemistä.
Jos olet kiihtynyt tai henkisesti rasittunut, tämä näkyy mittauksessa:
mitattu verenpaine on normaalia korkeampi (tai matalampi) ja pulssi on
yleensä tavallista nopeampi.
Yksilön verenpaine vaihtelee jatkuvasti riippuen siitä, mitä hän tekee tai
mitä hän on syönyt. Juomilla voi olla voimakas ja nopea vaikutus
verenpaineeseen.
Laitteen tekemät mittaukset pohjautuvat sydämen sykkeeseen. Jos syke
on erittäin heikko tai epäsäännöllinen, laitteen voi olla vaikea mitata
verenpainetta.
Jos laite huomaa epänormaalit olosuhteet, se keskeyttää mittauksen janäyttää virhesymbolin. Katso sivulta 6 symboleiden kuvaus.
Tämä verenpainemonitori on tarkoitettu aikuisten käyttöön. Kysy ohjeita
lääkäriltä ennen laitteen käyttöä lapsen verenpaineen mittaamiseen.
Lapset eivät saa käyttää tätä laitetta ilman valvontaa.
Liian korkeat lämpötilat tai suuri kosteus tai korkeus voivat vaikuttaa
automaattisen verenpainemonitorin toimintakykyyn.

TriCheckTM-mittaus
TriCheckTM-tila ottaa automaattisesti kolme peräkkäistä mittausta janäyttää näiden kolmen mittauksen keskiarvon.

TriCheckTM-tilan valitseminen
1.

Paina
paineenasetustilassa
SET-painiketta,
jotta
pääsetTM
TriCheck -asetustilaan. Nykyinen
asetus vilkkuu.

2.

Kytke TriCheckTM-tila päälle tai pois
- tai
päältä painamalla-painiketta.
PÄÄLLÄ: TriCheckTM-tila
POIS: Tavallinen mittaustila (oletusarvo)

3.

Sammuta laite painamalla STARTtai SET-painiketta . Laite sammuu
automaattisesti, jos sitä ei käytetä
kolmen minuutin kuluessa.

Suomi 16

Paineen asetustilassa:
paina

Nykyinen asetus vilkkuu
tai
Vaihda asetus
tai STARTMittaus TriCheckTM-tilassa
1.

2.

Paina START -painiketta.
Kaikki näytön alueet ovat näkyvissä.
Näytöllä näkyy hetken aikaa vilkkuva
nolla (0), ja ensimmäinen mittaus
alkaa.
Kun mittaus on valmis, systolisen ja
diastolisen paineen lukemat sekä
sydämenlyöntitiheys
tulevatnäyttöön. Tämän jälkeen käynnistyy
minuutin kestävä laskenta ennen
toista mittausta.

3.

Toinen mittaus alkaa minuutin
kuluttua.

4.

Kun mittaus on valmis, näytöllä
näkyy
ensimmäisen
ja
toisenmittauksen keskiarvolukema. Tämän
jälkeen käynnistyy minuutin kestävä
laskenta ennen kolmatta mittausta.

Sydämen tasolla

START

Paina
Kaikki näytön
alueet ovat näkyvissä
Ensimmäinen mittaus

5.

Kolmas mittaus alkaa minuutin
kuluttua.

6.

Kun mittaus on valmis, näytöllä näkyy
kolmen mittauksen keskiarvolukema,joka tallennetaan muistiin.

Huom.:
Mittauksen
aikananäkyy
TriCheckTM-symboli.
Voitperuuttaa
mittauksen
painamalla
START-painiketta.
Tässä tapauksessa mitään tietoja
ei tallenneta muistiin.
Jos
ensimmäisen
mittauksen
aikananäkyy
mansetinsopivuusvirheen symboli, peruuta
mittaus, aseta mansetti oikein ja
aloita uusi mittaus.
Mittauksen
jälkeenkolmenmittauksen keskiarvo tallennetaan
muistiin.
Mitään tietoja ei tallenneta, jos
mittaukset peruutetaan ennen
kolmannen mittauksen loppuun
suorittamista.
Suomi 17

Yhden minuutin jälkeen

Ensimmäisen
mittauksen tulokset

Toinen mittaus

Yhden minuutin jälkeen

Ensimmäisen ja toisen
mittauksen keskiarvo

Kolmas mittaus

Kolmen mittauksen
keskiarvo

Tietojen hakeminen muistista
Huom.: Laite tallentaa muistiin viimeiset 90 tietosarjaa automaattisesti.
1. Paina-tai -painiketta.
Paina
tai
Näyttöön tulee kaikkien mittausten
keskiarvo ja tiedon numero.
Systolinen keskiarvo(Jos tietoja ei ole, näyttöön tulee “0”.
Sammuta laite painamalla ,Diastolinen keskiarvo-taiSTART-painiketta.)
Pulssin keskiarvo
2. Muistin tiedot näytetään seuraavalla
Kaikkien tietojen-painikkeen (taitavalla jokaisella
keskiarvo-painikkeen, kun tiedot näytetään
käänteisessä järjestyksessä)
painalluksella.
Systolinen keskiarvo
Kaikkien aamulla (AM) klo 4:00–9:59
suoritettujen mittausten keskiarvo.
(Esimerkiksi 10 mittausta. Jos
tietoja ei ole, näyttöön tulee “--”.)

Diastolinen keskiarvo
Pulssin keskiarvo
AM-tietojen
keskiarvo

Kaikkien illalla (PM) klo 18:00–1:59
suoritettujen mittausten keskiarvo.
(Esimerkiksi 9 mittausta. Jos tietoja
ei ole, näyttöön tulee “--”.)
Viimeisimmät tiedot (nro n,esimerkissä nro 35)
Mittaustiedot näytetään kolme
sekuntia tietojen numeron
näyttämisestä.

Systolinen keskiarvo
Diastolinen keskiarvo
Pulssin keskiarvo
PM-tietojen
keskiarvo
Systolinen
Diastolinen

Viimeisimmät tiedot (nro 1)
Mittaustiedot näytetään kolme sekuntia
tietojen numeron näyttämisestä.

PulssiViimeisimmät tiedot

3. Kun viimeiset tiedot on näytetty, palaa
kaikkien tietojen keskiarvoa kuvaavalle-painiketta.
näytölle painamalla
4. Sammuta laite painamalla
START-painiketta. Laite sammuu
automaattisesti, jos sitä ei käytetä
minuutin kuluessa.

Suomi 18

Systolinen
Diastolinen
PulssiViimeisin tieto (vanhin)Mikä on IHB/AFib-indikaattori?
Jos monitori havaitsee epäsäännöllisen rytmin mittausten aikana,
IHB/AFib-indikaattori tulee näyttöön mittausarvojen kanssa.
Huom.: Kehotamme
ottamaanyhteyttä
lääkäriin,
jos
IHB/AFib-indikaattori näytetään toistuvasti.

Mikä on AFib?
Sydän supistuu sydämessä esiintyvien sähköisten signaalien vuoksi ja selähettää verta kaikkialle kehoon. Eteisvärinää (AFib) syntyy, kun
sähköinen signaali muuttuu eteisessä epäselväksi saaden aikaan
häiriöitä pulssirytmissä. AFib voi muodostaa sydämeen verihyytymiä.
Nämä voivat helposti synnyttää verisuonitukoksia, jotka puolestaan voivat
aiheuttaa aivohalvauksen ja sydänkohtauksen.

Paineen palkki-indikaattori
Indikaattori seuraa paineen muuttumista mittauksen aikana.
Paineistus
Ilmaa vapautetaan

Täyttö käynnissä

Mittaus käynnissä
Mansetti täynnä

WHO-luokitusindikaattori
Jokainen palkki-indikaattorin alue vastaa
WHO:n verenpaineluokitusta, joka on
kuvattu seuraavalla sivulla.
WHO-luokitusindikaattori
Erittäin korkea verenpaine
Melko korkea verenpaine
Lievästi kohonnut verenpaine
Korkea-normaali
Normaali
Ihanteellinen: Indikaattori näyttää, nykyisiin tietoihin
perustuen, alueen, joka vastaa WHO:nluokitusta.
Suomi 19

Esimerkki:

Melko korkea verenpaine

Lievästi kohonnut verenpaine

Korkea-normaali

Tietoja verenpaineesta
Mikä on verenpaine?

Verenpaine on voima, jolla veri painaa valtimoiden seinämää. Systolinen
paine syntyy, kun sydän supistuu. Diastolinen paine syntyy, kun sydän
laajenee. Verenpaine mitataan elohopeamillimetreinä (mmHg). Yksilön
luonnollista verenpainetta edustaa peruspaine, joka mitataan heti aamulla,
kun henkilö on vielä levossa, ennen aamiaista.

Mitä kohonnut verenpaine tarkoittaa ja kuinka sitä hoidetaan?

Kohonnut
verenpaine
tarkoittaa
epätavallisen
korkeaa
valtimoverenpainetta, joka hoitamattomana voi aiheuttaa monia
terveysongelmia, mukaan lukien aivohalvauksen tai sydänkohtauksen.
Kohonnutta verenpainetta voi hoitaa elämäntapamuutoksella, stressiä
välttämällä sekä lääkkeillä lääkärin valvonnassa.
Kohonneen verenpaineen ehkäiseminen tai hoitaminen:
Älä tupakoi
Harrasta säännöllistä liikuntaa
Vähennä suolaa ja rasvaa
Tarkista fyysinen kunto säännöllisesti
Ylläpidä hyvää painoindeksiä

Miksi verenpainetta kannattaa mitata kotona?

Verenpaineen mittaaminen sairaalassa tai lääkärin vastaanotolla voi
aiheuttaa hermostuneisuutta ja johtaa lukemiin, jotka ovat 25–30 mmHg
korkeammat kuin kotona mitatut. Kotona tehdyt mittaukset minimoivat
verenpainelukemiin vaikuttavat ulkoiset tekijät, täydentävät lääkäreiden
tekemiä mittauksia ja antavat tarkemman ja kattavamman kuvan
verenpaineen muutoksista ajan kuluessa.

WHO:n verenpaineluokitus

Verenpaineen vaihtelut

Diastolinen verenpaine

Maailmanterveysjärjestö Viitemateriaali: Journal of Hypertension 1999, Vol 17 No. 2
Luokan 3 kohonnut verenpaine (vakava)(WHO)onmäärittänyt mmHg
110standardit
korkean
105 Luokan 2 kohonnut verenpaine (kohtalainen)
verenpaineen
arvioinnille,
100
ikään katsomatta, kaaviossa
95 Luokan 1 kohonnut verenpaine (lievä)
kuvatun mukaisesti.
90
Korkea-normaali
85
Normaali
80 Ihanteellinen

Yksilön
verenpaine
vaihtelee
huomattavasti
120 130 140 150 160 170 180
Systolinen verenpaineverenpaineyksikössä mmHg
päivittäin ja kausittain. Painevoi vaihdella 30–50 mmHg
päivän eri tilanteissa. Henkilöillä, joilla on kohonnut verenpaine,
vaihtelut voivat olla jopa suurempia. Normaalisti verenpaine kohoaa
työnteon tai leikkimisen aikana, ja se laskee alimmilleen henkilön
nukkuessa. Tämän vuoksi yhdestä mittaustuloksesta ei kannata
huolestua liikaa.
Suomi 20

Tyypillinen vaihtelu päivän aikana(mitataan viiden minuutin välein)
Systolinen

Verenpaine

Voit selvittää normaalin verenpaineesi
tekemällä mittauksen joka päivä samaan
kellon aikaan tässä oppaassa kuvattuja
menetelmiä noudattaen. Säännöllisesti
tehdyt mittaukset antavat kattavamman
kuvan verenpaineen muutoksista ajan
kuluessa.
Muistamerkitäylösverenpainemittauksen päivämäärä jakellonaika. Pyydä lääkäriä tulkitsemaanverenpainetietoja.

Uni

Diastolinen
PM

AM

Kellonaika

Vianmääritys
Ongelma

Mahdollinen syy
Paristot ovat tyhjät.

Näytössä ei näy
mitään, vaikka Pariston navat eivät ole
laite on päällä. oikeilla paikoilla.

Mansetti ei
täyty.

Pariston jännite on liian
alhainen.
(PARISTO
VÄHISSÄ -merkki)
vilkkuu. Jos paristot
ovat tyhjentyneet täysin,merkki ei tule näkyviin.
Mansettia ei ole asetettu
oikein.
Käsivartta tai vartaloa
on liikutettu mittauksen
aikana.

Laite ei mittaa.
Lukemat ovat
liian korkeita tai Mansetti on väärässä
liian alhaisia.
kohdassa.

Muu

Arvo poikkeaa
sairaalassa tai lääkärin
vastaanotolla mitatusta
arvosta.

Suositeltu toimi
Vaihda kaikki paristot uusiin.
Asenna paristot uudelleen niin,
että negatiiviset ja positiiviset
navat on asetettu
paristolokerossa ilmoitetun
mukaisesti.

Vaihda kaikki paristot uusiin.

Laita mansetti oikein.
Varmista, että potilas pysyy
paikallaan ja puhumatta
mittauksen ajan.
Istu mukavasti ja liikkumatta.
Laita käsivarsi pöydälle niin,
että kämmen on ylöspäin ja
mansetti on sydämen tasolla.
Jos syke on erittäin heikko tai
epäsäännöllinen, laitteen voi ollavaikea mitata verenpainetta.
Katso kohta “Miksi
verenpainetta kannattaa mitata
kotona?”.

Poista paristot. Laita paristot
oikein takaisin ja mittaa
uudelleen.
Huom.: Jos yllä kuvatut toiminnot eivät ratkaise ongelmaa, ota yhteyttä
jälleenmyyjään. Älä yritä avata tai korjata tuotetta; se voisi mitätöidä
takuun.
Suomi 21

HuoltoÄlä avaa laitetta. Laitteessa on hienoja sähkökomponentteja ja
monimutkainen ilmayksikkö, jotka voisivat vahingoittua. Jos ongelmaa ei
voida ratkaista vianmääritysohjeita noudattamalla, ota yhteyttä alueesi
valtuutettuun jälleenmyyjään tai asiakaspalveluosastoomme. A&D:nasiakaspalvelu toimittaa valtuutetuille jälleenmyyjille teknisiä tietoja,
varaosia ja laitteita.
Laite on suunniteltu ja valmistettu niin, että sen käyttöikä on pitkä. Laiteon kuitenkin hyvä tarkistaa 2 vuoden välein sen asianmukaisen toiminnan
ja tarkkuuden varmistamiseksi. Ota yhteyttä alueesi valtuutettuun
jälleenmyyjään tai A&D:hen huollon tekemiseksi.

Tekniset tiedot
Tyyppi

UA-1020-W

Mittausmenetelmä

Oskillometrinen mittaus

Mittausalue

Paine:
0–299 mmHg
Systolinen paine: 60–279 mmHg
Diastolinen paine 40–200 mmHg
Pulssi:40–180 sykettä/minuutti

Mittaustarkkuus

Paine:
Pulssi:

Virtalähde

4 x 1,5 V paristot (R6P, LR6 tai AA) tai
Vaihtovirtasovitin (TB-233C) (ei sisälly toimitukseen)

Mittausten määrä

Noin 1 000 kertaa
LR6 (alkaliparistot)
Noin 300 kertaa
R6P (mangaaniparistot)
Painearvolla 180 mmHg huonelämpötilassa 23 °C.

Luokitus

Sähköinen lääkinnällinen laite sisäisellä
virtalähteellä (paristot) /Luokka II (sovitin)
Jatkuva toimintatila

Kliininen testi

Noudattaen standardia ISO81060-2：2013
Kliinisessä validointitutkimuksessa K5:ttä käytettiin
85 koehenkilön diastolisen verenpaineen
mittaamiseen.

EMD

IEC 60601-1-2: 2014

Muisti

90 viimeistä mittausta±3 mmHg±5 %Toimintaolosuhteet+10…+40 °C/suht. kosteus 15–85 %/800–1 060 hPa
Kuljetus-/säilytysolosuhteet -20…+60 °C /suht. kosteus 10–95 % / 700–1 060 hPa
Mitat
Noin 140 [L] x 60 [K] x 105 [S] mm
Suomi 22

PainoNoin 285 g, ilman paristoja

Liitäntäosa

Mansetti Tyyppi BF

Kotelointiluokka

Laite: IP20

Käyttöikä

Laite: 5 vuotta (kun käytetään kuusi kertaa päivässä)Mansetti: 2 vuotta (kun käytetään kuusi kertaa
päivässä)
Vaihtovirtasovitin: 5 vuotta (kun käytetään kuusi
kertaa päivässä)Lisävarusteena toimitettava
vaihtovirtasovitin
Sovittimella verenpainemonitori kytketään kodin
virtalähteeseen.
TB-233C

Sovittimen voi hankkia ottamalla yhteyttä
paikalliseen A&D-jälleenmyyjään.
Vaihtovirtasovitin pitää tarkistaa tai vaihtaa
säännöllisesti.
Vaihtovirtasovittimeen merkityt symbolit
Symbolit
Toiminta/Tarkoitus
Vain sisäkäyttöön
Luokan II laite
Lämpösulake
Sulake
EY:n direktiivin mukaisen laitteen merkintä
EAC-sertifioidun laitteen merkintä
Vaihtovirtasovittimen pistokkeen napaisuus

Lisävarusteet myydään erikseen
Mansetti
Kataloginumero
Mansetin koko
Käsivarren koko
CUF-F-LA
Suuri aikuisten mansetti
31 cm–45 cm
CUF-I
Laajan kokoalueen
22 cm–42 cm
mansetti
CUF-F-A
Aikuisten mansetti
22 cm–32 cm
CUF-F-SA
Pieni aikuisten mansetti
16–24 cm

Vaihtovirtasovitin
Kataloginumero
TB-233C

Pistoke
Tyyppi C

Huom.: Teknisiä tietoja voidaan muuttaa ilman ennakkoilmoitusta.
IP-luokitus on koteloiden antama suojausluokka standardin IEC 60529
mukaisesti. Laite on suojattu sellaisia vieraita esineitä vastaan, joiden
halkaisija on 12 mm tai suurempi, esim. sormi. Laitetta ei ole suojattu
vedeltä.
Suomi 23

MUISTIO

Suomi 24

Spis treści
Szanowny kliencie,................................................................ 2
Uwagi wstępne ..................................................................... 2Środki ostrożności ................................................................ 2
Oznaczenie części ................................................................ 4
Symbole ............................................................................... 5
Obsługa ................................................................................ 7
Korzystanie z ciśnieniomierza ............................................... 8
Wkładanie/wyjmowanie baterii ......................................... 8Podłączanie przewodu powietrza ..................................... 9Podłączanie zasilacza sieciowego ..................................... 9
Regulacja wbudowanego zegara .................................... 10
Wybór odpowiedniego mankietu .................................... 11
Zakładanie mankietu ...................................................... 11
Wykonywanie dokładnych pomiarów .............................. 13
Pomiar ........................................................................... 13
Po pomiarze .................................................................. 13
Pomiary .............................................................................. 14
Normalny pomiar ........................................................... 14
Pomiar z ustawionym ciśnieniem ................................... 15
Pomiar z określonym ciśnieniem skurczowym ................ 15
Uwagi dotyczące dokładnego pomiaru ........................... 16
Pomiar TriCheckTM .............................................................. 16
Wybieranie trybu TriCheckTM .......................................... 16
Pomiar za pomocą trybu TriCheckTM............................... 17
Wywoływanie zapisanych danych ........................................ 18
Czym jest wskaźnik IHB/AFib? ............................................ 19
Czym jest migotanie przedsionków (AFib)? ......................... 19Wskaźnik słupkowy ciśnienia .............................................. 19Wskaźnik klasyfikacji WHO .................................................. 19
Informacje o ciśnieniu tętniczym......................................... 20
Czym jest ciśnienie tętnicze?.......................................... 20
Czym jest nadciśnienie i jak można je kontrolować? ....... 20
Dlaczego warto mierzyć ciśnienie tętnicze w warunkachdomowych? ........................................................................ 20
Klasyfikacja ciśnienia tętniczego wg WHO ...................... 20
Zmiany ciśnienia tętniczego ........................................... 20
Rozwiązywanie problemów ................................................. 21
Konserwacja ....................................................................... 22
Dane techniczne ................................................................. 22
Polski 1

Szanowny kliencie,
gratulujemy zakupu ciśnieniomierza A&D. Łatwość obsługi i wysoka
dokładność ciśnieniomierza pozwolą na skuteczne monitorowanie ciśnieniatętniczego.
Przed pierwszym użyciem urządzenia zalecamy dokładne przeczytanie
niniejszej instrukcji.

Uwagi wstępne
Niniejsze urządzenie spełnia wymagania Dyrektywy Rady 93/42 EWG dotyczącej
wyrobów medycznych. Jest to potwierdzone znakiem zgodności
.
(0123: numer referencyjny jednostki notyfikowanej)Urządzenie jest przeznaczone do użycia przez osoby dorosłe — zabrania się
dokonywania pomiarów ciśnienia u noworodków i niemowląt.
Środowisko użytkowania. Urządzenie jest przeznaczone do obsługi samodzielnej
w warunkach domowych.
Niniejsze urządzenie jest przeznaczone do pomiarów ciśnienia tętniczegoi tętna dla celów diagnostycznych.

Środki ostrożności
Do konstrukcji tego urządzenia użyto precyzyjnych podzespołów. Należy
chronić je przed działaniem skrajnych temperatur, wilgoci, bezpośredniegoświatła słonecznego, uderzeń oraz kurzu.
Urządzenie lub mankiet należy czyścić miękką, suchą ściereczką lubściereczką nasączoną wodą i neutralnym detergentem. Do czyszczenia nigdy
nie używać alkoholu, benzenu, rozcieńczalników ani innych środków
chemicznych o agresywnym działaniu.
Nie należy składać mankietu zbyt ciasno ani nie należy przechowywać
przewodu mocno skręconego przez dłuższy czas, gdyż może to spowodować
zmniejszenie trwałości tych elementów.
Należy uważać, by nie doszło do przypadkowego uduszenia się niemowlęcia
lub dziecka na skutek kontaktu z przewodem.
W trakcie pomiaru nie należy skręcać przewodu. Może to być przyczyną
powstania obrażeń, ponieważ mankiet znajduje się stale pod ciśnieniem.
Urządzenie oraz mankiet nie są wodoodporne. Unikać kontaktu urządzenia i
mankietu z deszczem, potem i wodą.
W przypadku korzystania z urządzenia w pobliżu telewizorów, kuchenek
mikrofalowych, aparatów rentgenowskich i innych urządzeń o silnym polu
elektrycznym, wyniki pomiaru mogą być zaburzone.
Na działanie monitora ciśnienia tętniczego mogą mieć także wpływ urządzenia
komunikacyjne, takie jak urządzenia działające w sieci domowej, telefony
komórkowe, telefony bezprzewodowe i ich stacje bazowe oraz krótkofalówki.
Zalecane jest utrzymanie odległości ok. 30 cm od takich urządzeń.
Zużyte urządzenie, części oraz baterie nie mogą być traktowane jak normalne
odpady z gospodarstwa domowego i należy je zutylizować zgodnie
z obowiązującymi lokalnymi przepisami.
W przypadku korzystania z zasilacza sieciowego należy upewnić się, że w razie
konieczności będzie można go w łatwy sposób wyjąć z gniazda sieciowego.
Przed ponownym użyciem urządzenia należy upewnić się, że jest czyste.
Polski 2Urządzenia nie należy poddawać modyfikacjom. Może to stać się przyczyną
wypadków lub doprowadzić do uszkodzenia urządzenia.
Aby dokonać pomiaru ciśnienia tętniczego, należy na tyle mocno ścisnąć ramię
mankietem, by chwilowo zatrzymać przepływ krwi przez tętnicę. Może tospowodować ból, drętwienie lub tymczasowe zaczerwienienie ramienia. Takistan może pojawiać się zwłaszcza w sytuacjach, gdy pomiar jest kolejno
powtarzany.
W miarę upływu czasu objawy zanikają.
Zbyt częste pomiary ciśnienia tętniczego mogą być szkodliwe z uwagi na
zakłócenie przepływu krwi. Należy upewnić się, że wielokrotne używanieurządzenia nie będzie skutkowało przedłużonym zaburzeniem przepływu krwi.
Po zabiegu mastektomii przed użyciem urządzenia należy skontaktować się zlekarzem.
Nie wolno pozwalać dzieciom na samodzielne korzystanie z urządzenia ani nienależy używać urządzenia w pobliżu niemowląt. Może to stać się przyczyną
wypadków lub doprowadzić do uszkodzenia mienia.
Urządzenie zawiera małe części, które w przypadku połknięcia przez
niemowlęta stwarzają ryzyko udławienia się.
Jeśli zasilacz nie będzie używany podczas dokonywania pomiaru, należy wyjąć
go z gniazda sieciowego.
Używanie akcesoriów niewymienionych w niniejszej instrukcji może obniżyć
poziom bezpieczeństwa.
W przypadku zwarcia baterii może ona nagrzać się do wysokiej temperatury,
stwarzając zagrożenie poparzeniem.
Przed użyciem należy pozostawić urządzenie na około godzinę, aby umożliwić
jego dostosowanie się do warunków otoczenia.
Nie przeprowadzono badań klinicznych u noworodków ani kobiet w ciąży.
Urządzenia nie należy używać do pomiaru ciśnienia krwi u noworodków i kobiet
w ciąży.
Nie należy dotykać jednocześnie baterii, gniazda DC i ciała pacjenta. Może tospowodować porażenie prądem elektrycznym.
Nie nadmuchiwać mankietu niezałożonego na ramię.

PrzeciwwskazaniaPoniżej opisano środki ostrożności dotyczące właściwego korzystania z urządzenia.
Nie zakładać mankietu na ramię, na które założono inne urządzenia medyczne.
Urządzenia te mogą nie funkcjonować prawidłowo.
Osoby, które mają poważne problemy krążeniowe w ramieniu powinny przedużyciem urządzenia skonsultować się z lekarzem, aby zapobiec problemomnatury medycznej.
Nie należy samodzielnie dokonywać diagnozy ani podejmować leczenia
w oparciu o uzyskane wyniki. W celu oceny wyników i wyboru sposobu
leczenia należy zawsze konsultować się z lekarzem.
Nie zakładać mankietu na ramię, na którym znajduje się niezagojona rana.
Nie zakładać mankietu na ramię, do którego podłączony jest wlew dożylny lub
do którego dokonywana jest transfuzja krwi. Może to stać się przyczyną
odniesienia obrażeń lub wypadków.
Nie używać urządzenia w obecności łatwopalnych gazów jak np. gazznieczulający. Może być to przyczyną wybuchu.
Nie używać urządzenia w warunkach wysokiego stężenia tlenu, takich jak
wysokociśnieniowe komory tlenowe lub namioty tlenowe. Może to stać się
przyczyną pożaru lub wybuchu.
Polski 3

Oznaczenie części
Gniazdo DCWyświetlacz

Przycisk
Przycisk
Przycisk

SET

Wtyk przewodu powietrza
Przycisk START
Gniazdo przewodu
powietrza
Przewód powietrza
Prawidłowy zakres
dopasowaniaWskaźnik pozycji tętnicy
Mankiet naramienny

Znacznik
Pokrywa komory baterii
Komora baterii

Baterie 1,5 V(R6P, LR6 lub AA)Wyświetlacz

PAMIĘĆ

Wartość uśredniona

Ustawienia ciśnienia

Symbol IHB/AFib
Symbol nieprawidłowego dopaso
wania mankietuCiśnienie
SkurczoweWskaźnik błędu ruchu
Symbol TriCheckTMCiśnienie
rozkurczoweWskaźnik sercaWskaźnik klasyfikacji WHO
i wskaźnik słupkowy ciśnieniaTętnoWskaźnik poziomu naładowania baterii
Symbol AM/PM
Polski 4

Godziny

Symbole
Symbole na obudowie urządzenia i zasilaczu sieciowym
SymboleSN

Funkcja/znaczenie

Zalecane działanie

Włączanie i wyłączanie urządzenia.
Wskazówka dotycząca montażu bateriiPrąd stały
Numer seryjny
Data produkcji
Typ BF: urządzenie, mankiet
i przewody zostały zaprojektowane tak,aby zapewnić specjalną ochronę przedporażeniem prądem elektrycznym.
Etykieta dyrektywy WE odnośnie
urządzeń medycznych
Etykieta WEEE
ProducentPrzedstawiciel w UE
Polaryzacja gniazda DC

IPMiędzynarodowy
symbolochronyurządzenia
Przechowywać w suchym miejscu

Symbole widoczne na wyświetlaczu
SymboleFunkcja/znaczenie
Symbol widoczny podczas pomiaru.
Zaczyna migać po wykryciu tętna.
Symbol IHB/AFib
Pojawia się, gdy zostaje wykryte
nieregularne bicie serca.
Może wyświetlić się, gdy zostanie
wykryta bardzo łagodna wibracja jak
np. drżenie lub dreszcze.

Zalecane działanie
Pomiar w toku. Unikać
ruchów.

Tryb TriCheckTM
Automatyczne pobieranie trzech
kolejnych pomiarów i wyświetlanieuśrednionych wartości tych pomiarów.
Wyświetla się po wykryciu ruchu ciała
lub ramienia.
Polski 5

Pomiar może być
niedokładny. Należy
wykonać kolejny pomiar.
Pozostawać w bezruchu.

Symbole

Funkcja/znaczenie
Pojawia się podczas pomiaru, jeśli
mankiet nie zostanie dokładniezałożony.

Zalecane działanie
Pomiar może być
niedokładny. Założyć
mankiet w prawidłowy
sposób i ponowić pomiar.

Wyniki poprzednich pomiarów zapisanew pamięci.
Wartość uśredniona danych
BATERIA NAŁADOWANA
Wskaźnik poziomu naładowania baterii
podczas pomiaru.
NISKI POZIOM BATERIIMiganie oznacza niski stopień
naładowania baterii.
Niestabilne ciśnienie tętnicze
spowodowane poruszaniem się
podczas pomiaru.
Różnica pomiędzy wartością
skurczową a rozkurczową zawiera się
w przedziale 10 mmHg.
Wartość ciśnienia podczas
nadmuchiwania nie wzrasta.
Mankiet nie został zapięty prawidłowo.

Gdy ten symbol będzie
migać, należy wymienić
baterie.
Zaleca się ponowić
pomiar.
Pozostawać w bezruchu.

Zamocować mankiet
prawidłowo i ponowić
pomiar.

BŁĄD TĘTNA
Nie wykryto prawidłowego tętna.

Wewnętrzny błąd monitora ciśnieniatętniczego

SYS

Skurczowe ciśnienie tętnicze w mmHg.

DIA

Rozkurczowe ciśnienie tętnicze
w mmHg.

PUL

Uderzenia na minutę

AM
PM

Dane pobrane w okresie od godziny
4:00 do 9:59
Dane pobrane w okresie od godziny
18:00 do 1:59
Ustawienia ciśnienia
Wskazuje wartość ciśnienia ustawioną
wcześniej przez użytkownika.
Polski 6Wyjąć baterie i nacisnąć
przycisk START , anastępnie ponownieumieścić bateriew urządzeniu. Jeżeli błądbędzie występował nadal,należy skontaktować się ze
sprzedawcą.

Obsługa
1. Normalny pomiar
Nacisnąć przycisk START . Ciśnienie zostanie zmierzone, a dane zostaną
zapisanew urządzeniu.
Urządzeniemożeautomatycznie
zapisaćdziewięćdziesiąt zestawów danych w pamięci.

2. Wyświetlanie zapisanych danych
Wcisnąć przycisk
lub , aby wyświetlić dane
zapisane w pamięci. Wyświetlona zostanie
wartość średnia wszystkich pomiarów, jak
przedstawiono na ilustracji po prawej stronie.
Następnie każdorazowe naciśnięcie przycisku
spowoduje wyświetlenie danych zapisanych
w pamięci w sposób określony poniżej.
Wartość uśredniona wszystkich pomiarów
AM (porannych), przeprowadzonych w
okresie od godziny 4:00 do 9:59.
Wartość uśredniona wszystkich pomiarów
PM (wieczornych), przeprowadzonych w
okresie od godziny 18:00 do 1:59.
Najnowsze dane (Nr, na przykładzie Nr 35)
Ostatnie dane (Nr. 1)
Szczegółowe
informacje
dotyczące
przywoływania
zapisanych
danychzamieszczono
w części
„Wyświetlanie
zapisanych danych”. (Zob. str. 18.)

3. Usuwanie wszystkich danych z pamięci
Wcisnąć oba przyciskii . Pojawią się symbol
oraz wskaźnik poziomu naładowania baterii. Wcisnąć i
przytrzymać oba przyciskioraz
do momentu, ażzacznie migać, aby usunąć
podświetlony symbol
wszystkie dane przechowywane w pamięci.

4. Pomiar z określonym ciśnieniem skurczowym
Informacje na temat pomiaru
zamieszczono na stronie 15.

z określonym

Polski 7ciśnieniem

skurczowym

Korzystanie z ciśnieniomierza
Wkładanie/wyjmowanie baterii
1. Zdjąć pokrywę komory baterii.
2. Wyjąć zużyte baterie i włożyć w ichmiejsce nowe baterie do komory
w przedstawiony sposób, zwracając
uwagę na polaryzację (+ i -).
Należy stosować tylko baterie R6P,
LR6 lub AA.
3. Założyć pokrywę komory baterii.

Krok 1
Krok 3

Krok 1
Krok 2OSTROŻNIE
Baterie należy umieszczać w komorze w sposób przedstawiony na ilustracji.
W przypadku nieprawidłowego włożenia baterii urządzenie nie włączy się.
Gdy na wyświetlaczu zacznie migać symbol(NISKI POZIOM BATERII),należy wymienić wszystkie baterie na nowe. Nie używać jednocześnie bateriizużytych i nowych. Może to spowodować skrócenie ich żywotności lub
nieprawidłowe działanie urządzenia.
Wymienić baterie po upływie co najmniej dwóch sekund od wyłączenia sięurządzenia.
(NISKI POZIOM BATERII) będzie nadal widoczny, należy
Jeśli symbol
wykonać pomiar ciśnienia tętniczego. Podczas pomiaru urządzenie powinno
wykryć stan naładowania baterii.
(Symbol NISKI POZIOM BATERII) nie będzie wyświetlany, jeśli baterieulegną całkowitemu rozładowaniu.
Żywotność baterii zależy od temperatury otoczenia i może być mniejsza
w niższych temperaturach. Zasadniczo cztery nowe baterie LR6 powinny
wystarczyć na ok. rok w przypadku codziennych pomiarów dokonywanych dwa
razy dziennie.
Należy używać wyłącznie baterii wymienionych w instrukcji. Baterie dostarczone
wraz z urządzeniem służą wyłącznie do sprawdzenia działania urządzenia imogą mieć ograniczoną żywotność.
Jeżeli urządzenie nie będzie używane przez dłuższy czas, należy wyjąć baterie.
Baterie mogą wyciec i spowodować awarię urządzenia.

Polski 8

Korzystanie z ciśnieniomierzaPodłączanie przewodu powietrza
Dokładnie włożyć wtyk przewodu
powietrza w gniazdo przewodu
powietrza.

Gniazdo przewodu
powietrza

Wtyk przewodu powietrzaPodłączanie zasilacza sieciowegoWłożyć wtyczkę zasilacza do gniazdaDC.
Następnie podłączyć zasilacz dogniazda sieciowego.

Wtyk zasilacza sieciowego

Stosować tylko zasilacz opisany w instrukcji.
Gniazdo DC(Zob. str. 23.)Odłączając zasilacz od gniazda sieciowego, należy chwycić obudowę
zasilacza i wyjąć go z gniazda.
Odłączając wtyczkę zasilacza od monitora ciśnienia tętniczego, należy
chwycić i wyjąć wtyczkę zasilacza z urządzenia.

Polski 9

Korzystanie z ciśnieniomierza
Regulacja wbudowanego zegara
Wyregulować zegar przed skorzystaniem z urządzenia.
1. Naciskać przycisk

SET , aż rok zacznie migać.

lub .
2. Wybrać rok za pomocą przycisku
Wcisnąć przycisk SET , aby zapisać aktualny
rok i przejść do opcji wyboru miesiąca/dnia.
Istnieje możliwość ustawienia dowolnej daty z
lat 2010–2059.

Nacisnąć

Roklub

Miesiąc

3. Wybrać miesiąc za pomocą przycisku
lub .
Wcisnąć przycisk SET , aby zapisać aktualny
miesiąc i przejść do opcji wyboru dnia.
lub .
4. Wybrać dzień za pomocą przycisku
Wcisnąć przycisk SET , aby zapisać aktualnydzień i przejść do opcji wyboru godziny/minut.
5. Wybrać godzinę za pomocą przycisku
lub .
Wcisnąć przycisk SET , aby zapisać aktualną
godzinę i przejść do opcji wyboru minut.
lub .
6. Wybrać minuty za pomocą przycisku(Wcisnąć przycisk SET , aby przejść do trybu
ustawiania ciśnienia. Zapoznać się ze
szczegółami podanymi na stronie 15).
Nacisnąć przycisk START, aby wyłączyć
urządzenie.

lub

Dzień

lub

lub

Godz.

Min.

lub

Uwaga: Po trzech minutach bezczynności urządzeniewyłączy się automatycznie.
Jeżeli nie ustawiono zegara,Wyłączenie
zamiast godziny wyświetlaneNaciśnięcie przycisku Trybbędą kreski, jak pokazano na
START w dowolnym ustawiania ciśnienia
prawej stronie.
momencie spowoduje
Przytrzymanie przycisku
lubwyłączenie urządzenia.
spowoduje ciągłą zmianę wartości.

Polski 10

Korzystanie z ciśnieniomierza
Wybór odpowiedniego mankietuUżywanie mankietu w odpowiednim rozmiarze jest istotne dla dokładnych
odczytów. Niewłaściwy mankiet może spowodować nieprawidłowy odczyt
ciśnienia tętniczego.
Rozmiar ramienia nadrukowano na każdym mankiecie.
Znacznikoraz zakres dopasowania widoczne na mankiecie pozwolą określić,czy wybrano prawidłowy rozmiar mankietu. (Zob. tabelę „Symbole na mankiecie”
na następnej stronie).
Jeśli znacznik
wskazuje wartość poza zakresem, należy skontaktować sięz lokalnym dystrybutorem w celu zakupu innego mankietu.
Mankiet podlega zużyciu. W przypadku jego zużycia należy zakupić nowy
mankiet.
Obwód ramienia
Zalecany rozmiar mankietu
Od 31 cm do 45 cm
Duży mankiet dla dorosłych
Od 22 cm do 42 cm Mankiet o zwiększonym zakresie
Od 22 cm do 32 cm
Mankiet dla dorosłych
Od 16 cm do 24 cm
Mały mankiet dla dorosłych
Obwód ramienia: obwód mierzony przy bicepsie.

Numer katalogowy
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

Zakładanie mankietu
1. Owinąć ramię mankietem, około 1–2
cm powyżej wewnętrznej strony łokcia,
jak pokazano na ilustracji.
Mankiet musi znajdować się
bezpośrednio na skórze, gdyż ubraniemoże zafałszować odczyt ispowodować błędny pomiar.

Nie należy podwijać rękawa zbyt ciasno.

Mankiet1–2 cmPrzewód powietrzaWskaźnik pozycji tętnicy

2. Zwężenie ramienia spowodowane
zawiniętym
rękawemmożeprzeszkodzić w przeprowadzeniu
dokładnego pomiaru.

Zabezpieczyć
mocowanie z tkaniny.

3. Upewnić się, że znacznik
wskazuje odpowiedni zakres.
Znacznik

Polski 11

i zakres dopasowania

Korzystanie z ciśnieniomierza
Symbole na mankiecie
SymboleFunkcja/znaczenie

Zalecane działanie
Umieścić wskaźnik
na
wysokości tętnicy ramienia
lub w jednej linii z palcem
serdecznympowewnętrznej stronie ręki.

Wskaźnik pozycji tętnicy▲
REF

Znacznik
Numer katalogowy
Prawidłowy zakres dopasowania dla
mankietu dla dorosłych.
Nadrukowany na mankiecie dla dorosłych.
Użyć dużego mankietu dla
Zakres nadmiarowy nadrukowany na dorosłych zamiast mankietu
mankiecie dla dorosłych i mankiecie
dla dorosłych lub mankietu
o zwiększonym zakresie.
o zwiększonym zakresie.
Prawidłowy
zakresdopasowania
mankietuo zwiększonym
zakresie.
Nadrukowany
namankiecieo zwiększonym zakresie.
Prawidłowy zakres dopasowania dużego
mankietu dla dorosłych. Nadrukowany nadużym mankiecie dla dorosłych.
Zakres pomniejszony nadrukowany na Użyć małego mankietu dlazamiast mankietu
mankiecie dla dorosłych i mankiecie dorosłych
dla dorosłych lub mankietu
o zwiększonym zakresie.
o zwiększonym zakresie.
Prawidłowy zakres dopasowania dlamałegomankietudladorosłych.
Nadrukowany na małym mankiecie dla
dorosłych.
Użyć mankietu dla dorosłych
Nadrukowany na dużym i małym zamiast dużego lub małegomankiecie dla dorosłych.
mankietu dla dorosłych.
Numer partii
Prawidłowy zakres dopasowaniaDuży mankiet dla
dorosłychA

L

Mankiet o zwiększonym
zakresie

S

W

L

Mankiet dla dorosłych

S

A

L

S

A

Mały mankiet dla dorosłych
Polski 12

Korzystanie z ciśnieniomierza
Wykonywanie dokładnych pomiarów
Aby uzyskać najdokładniejszy pomiar ciśnienia tętniczego, należy:Usiąść wygodnie na krześle. Oprzeć rękę na stole. Nie krzyżować nóg. Oprzeć
stopy na podłodze i wyprostować się.
Przed pomiarem zaleca się odprężyć przez od pięciu do dziesięciu minut.
Umieścić środkową część mankietu na poziomie serca.
Nie poruszać się ani nie rozmawiać podczas pomiaru.
Nie dokonywać pomiarów bezpośrednio po ćwiczeniach fizycznych lub po kąpieli.
Przed pomiarem należy odpocząć od dwudziestu do trzydziestu minut.
Pomiary ciśnienia tętniczego należy wykonywać w miarę możliwości codziennie o
tej samej porze dnia.

Pomiar
Podczas pomiaru mankiet może uciskać ramię — jest to zjawisko normalne.
Zachować spokój.

Po pomiarze
Po pomiarze nacisnąć przycisk START , aby wyłączyć urządzenie.
Zdjąć mankiet i zapisać dane. Po minucie bezczynności urządzenie wyłączy się
automatycznie.

Polski 13

Pomiary
Przed pomiarem należy przeczytać punkt „Uwagi dotyczące dokładnego
pomiaru” na stronie 16.

Normalny pomiar
1.

Założyć mankiet na ramię (zaleca się założenie na
lewe ramię).
Podczas pomiaru nie rozmawiać.

2.

Nacisnąć przycisk START .
Wyświetlą
się
wszystkie
segmentywyświetlacza.
Wyświetli się cyfra zero (0) i zacznie migać.
Wyświetlacz zmieni się, wskazując dane
przedstawione na ilustracji po prawej stronie,
a pomiar zostanie rozpoczęty. Ciśnienie
w mankiecie zacznie rosnąć. Mankiet możebardzo uciskać ramię — jest to zjawisko
normalne. Podczas nadmuchiwania po lewej
stronie wyświetlacza pojawi się pasek ciśnienia.
Uwaga: abyzatrzymać
nadmuchiwanie
mankietu w dowolnej chwili, należy
ponownie nacisnąć przycisk START .

3.

Po napompowaniu mankietu powietrze
zostanie automatycznie spuszczone, a symbol(serce) zacznie migać, wskazując trwanie
pomiaru. Po wykryciu tętna znacznik zacznie
migać zgodnie z tętnem.
Uwaga: jeśli odpowiednia wartość ciśnienia nie
zostanie uzyskana, urządzenie zacznie
automatycznie nadmuchiwać mankiet.
Aby uniknąć konieczności
ponownego nadmuchiwania, należy
zapoznać się z punktem „Pomiar
z ustawionym ciśnieniem” nanastępnej stronie.

4.

Po zakończeniu pomiaru wyświetlą się
wartości
ciśnienia
skurczowego
i rozkurczowego oraz tętna.
Z mankietu zostanie usunięte powietrze.

5.

Nacisnąć przycisk START , aby wyłączyć
urządzenie.
Po minucie bezczynności urządzeniewyłączy się automatycznie.

Uwaga: w razie wykonywania pomiaru na tej
samej osobie należy odczekać conajmniej trzy minuty między kolejnymi
pomiarami.
Polski 14

Na wysokości
serca
Nacisnąć

START

Wyświetlenie
wszystkich
segmentówwyświetlacza

Wskazanie zerowe
Rozpoczęcie
nadmuchiwania

Nadmuchiwanie

Pomiarw tokuCiśnienie skurczoweCiśnienie rozkurczowe
Klasyfikacja WHOTętnoAutomatyczne
usunięcie pozostałego
powietrza

PomiaryUrządzenie UA-1020-W pozwala na pomiar tętna oraz automatyczne
nadmuchanie mankietu do poziomu ciśnienia skurczowego.
Jeżeli mankiet jest ciągle ponownie nadmuchiwany, należy zastosować
poniższe metody.

Pomiar z ustawionym ciśnieniem

Podczas pomiaru ciśnienia tętniczego mankiet może zostać nadmuchany
ponownie.
Aby uniknąć ponownego nadmuchiwania, można ustawić odpowiednią wartość
ciśnienia.

1.

W kroku 6 procedury regulacji zegara (patrz
strona 10) wcisnąć przycisk SET , aby przejść
do trybu ustawiania ciśnienia. Aktualne
ustawienie zacznie migać.

2.

Wcisnąć przycisk
lub, aby wybrać
wartość ciśnienia wyższą o przynajmniej około
30 mmHg od oczekiwanego ciśnienia
skurczowego w poniższych ustawieniach.
AUTO: Automatyczne ustawianie ciśnienia(wartość domyślna)180: Ciśnienie o wartości 180 mmHg (stałe)
210: Ciśnienie o wartości 210 mmHg (stałe)
240: Ciśnienie o wartości 240 mmHg (stałe)

3.

lub

Wyłączenie

Tryb ustawiania
TriCheck

Wcisnąć przycisk SET , aby przejść do trybu ustawiania TriCheck.
Nacisnąć przycisk START , aby wyłączyć urządzenie. Po trzech minutach
bezczynności urządzenie wyłączy się automatycznie.
Kolejny pomiar zostanie przeprowadzony z nową wartością ciśnienia.

Pomiar z określonym ciśnieniem skurczowymNależy skorzystać z tej metody gdy mankiet zostanie nadmuchany kilkukrotnie
pod rząd, nawet jeśli w ramach powyższej procedury ustawiono ciśnienie na
poziomie 240 lub gdy wyniki pomiaru nie wyświetlą się, nawet jeśli ciśnienie
spadnie do wartości 20 mmHg lub niższej.

1.

Założyć mankiet na ramię (zaleca sięzałożenie na lewe ramię).

Na wysokości serca

2.

Nacisnąć i przytrzymać przycisk START ,aż pojawi się wartość wyższa od
spodziewanego ciśnienia skurczowego o
30 do 40 mmHg.

Nacisnąć i przytrzymać START
przycisk3.

Gdy zostanie osiągnięta żądana wartość,puścić przycisk START , aby rozpocząć
pomiar. Kontynuować pomiar ciśnieniatętniczego
w sposób
opisanyna
poprzedniej stronie.

Polski 15

Zwolnić przycisk poosiągnięciu żądanej
wartości ciśnienia
skurczowego

Informacje dotyczące pomiaru
zamieszczono na poprzedniej stronie.

Uwagi dotyczące dokładnego pomiaruNależy usiąść w wygodnej pozycji. Oprzeć rękę na stole z dłonią skierowaną ku
górze i mankietem założonym na wysokości serca.
Przed rozpoczęciem pomiaru zaleca odprężyć się przez 5-10 minut. U osób
podekscytowanych lub przygnębionych w wyniku stresu emocjonalnego pomiar
odzwierciedli te stany jako wyższe (lub niższe) ciśnienie tętnicze niż w stanie
normalnym, a wartość tętna będzie zwykle wyższa niż normalnie.
Ciśnienie tętnicze krwi danej osoby podlega ciągłym zmianom, w zależności odwykonywanej czynności oraz spożytych pokarmów. Przyjmowane napoje mogąistotnie i gwałtownie wpływać na ciśnienie tętnicze.
Urządzenie opiera swój pomiar na tętnie. Jeżeli tętno jest bardzo słabe lub
nieregularne, urządzenie może mieć problemy z określeniem ciśnienia tętniczego.
Jeżeli urządzenie wykryje stan odbiegający od normy, wówczas pomiar
zostanie przerwany i wyświetli się komunikat o błędzie. Opis symboli
zamieszczono na stronie 6.
Ten monitor ciśnienia tętniczego jest przeznaczony wyłącznie dla osób
dorosłych. Przed zastosowaniem urządzenia u dziecka należy skonsultowaćsię z lekarzem. Dzieci nie mogą używać urządzenia bez nadzoru.
Na działanie automatycznego monitora ciśnienia tętniczego może wpływać
nadmierna temperatura, wilgotność lub wysokość nad poziomem morza.

TM

Pomiar TriCheck

Tryb TriCheckTM polega na automatycznym pobieraniu trzech kolejnych
pomiarów i wyświetlaniu uśrednionych wartości tych pomiarów.

Wybieranie trybu TriCheckTM

W trybie ustawiania ciśnienia,

1.

W trybie ustawiania ciśnienia wcisnąć
przycisk SET , aby przejść do trybu
ustawiania
TriCheckTM.
Aktualne
ustawienie zacznie migać.

2.

lub
, aby
Nacisnąć przycisk
włączyć lub wyłączyć tryb TriCheckTM.
Aktualne ustawienie zacznie migać.
WŁ.: Tryb TriCheckTM
WYŁ.: Normalny tryb pomiaru (wartość domyślna)nacisnąć

lub

3.

Wcisnąć przycisk START lub SET ,aby wyłączyć urządzenie. Po trzech
minutach bezczynności urządzeniewyłączy się automatycznie.

Polski 16

Zmienić ustawienie
lub STARTPomiar za pomocą trybu TriCheckTM
1.

2.

Nacisnąć przycisk START .
Wyświetlą się wszystkie segmentywyświetlacza.
Wyświetli się cyfra zero (0), która będzie
przez chwilę mrugać i rozpocznie siępierwszy pomiar.
Po zakończeniu pomiaru wyświetlą się
wartości
ciśnienia
skurczowego
i rozkurczowego oraz tętna, po czym
rozpocznie się odliczanie jednej minuty,
po którym rozpoczęty zostanie drugi
pomiar.

3.

Po upływie jednej minuty rozpocznie się
drugi pomiar.

4.

Po zakończeniu pomiaru wyświetlą sięuśrednione wartości pierwszego idrugiego pomiaru, po czym rozpoczniesię odliczanie jednej minuty, po którym
rozpoczęty zostanie trzeci pomiar.

5.
6.

Po upływie jednej minuty rozpocznie siętrzeci pomiar.

Na wysokości
serca
START

Nacisnąć
Wyświetlenie
wszystkich
segmentówwyświetlacza
Pierwszy pomiar

Po jednej minucie

Wynikipierwszego pomiaru

Drugi pomiar

Po zakończeniu pomiaru wyświetlą sięuśrednione wartości trzech pomiarów,
które zostaną zapisane w pamięci.

Uwagi:
Podczas pomiaru wyświetlany będzie
symbol TriCheckTM.
Nacisnąć przycisk
START , aby
anulowaćpomiar.
Wtym
przypadku dane nie zostaną zapisanew pamięci.
Gdy podczas pierwszego pomiaru
pojawi się symbol nieprawidłowego
dopasowania
mankietu,należyanulować ten pomiar, prawidłowozałożyć mankiet oraz rozpocząć nowy
pomiar.
Po zakończeniu pomiaru daneuśrednione trzech pomiarów zostaną
zapisane w pamięci.
Anulowanie
pomiarów
przedzakończeniem trzeciego pomiaru
spowoduje, że żadne dane nie
zostaną zapisane.
Polski 17

Po jednej minucieUśredniona wartość
pierwszego idrugiego pomiaru

Trzeci pomiar

Wartość uśredniona
trzech pomiarów

Wywoływanie zapisanych danych
Uwaga: Urządzenie automatycznie zapisuje ostatnie dziewięćdziesiąt zestawów
danych w pamięci.
1. Nacisnąć przycisk
lub .
Nacisnąć
lubWyświetlą się uśredniona wartość ze
wszystkich pomiarów oraz liczba
zapisanych danych.
Średnie ciśnienie skurczowe(W razie braku zapisanych danychŚrednie ciśnienie rozkurczowewyświetli się cyfra „0”. Nacisnąć przycisk ,lub START, aby wyłączyć urządzenie).
Średnie tętno
2. Każdorazowe naciśnięcie przycisku
Wartość uśredniona(lub przycisku
do wyświetlania
wszystkich danych
danych w odwrotnej kolejności)spowoduje wyświetlenie danych
zapisanych w pamięci w sposóbokreślony poniżej.
Średnie ciśnienie skurczowe
Wartość
uśredniona
wszystkichŚrednie ciśnienie rozkurczowe
pomiarów
AM(porannych),Średnie tętno
przeprowadzonych w okresie odgodziny 4:00 do 9:59.
Wartość uśredniona(10 pomiarów na przykładzie. W razie
danych AM
braku zapisanych danych wyświetli się
symbol „--”.)
Wartość
uśredniona
wszystkich
pomiarówPM(wieczornych),przeprowadzonych w okresie odgodziny 18:00 do 1:59.
(9 pomiarów na przykładzie. W razie
braku zapisanych danych wyświetli się
symbol „--”.)
Najnowsze dane (Nr, na przykładzie
Nr 35)
Dane pomiarowe zostaną wyświetlone
trzy sekundy po wyświetleniu numeru
danych.

Średnie ciśnienie skurczoweŚrednie ciśnienie rozkurczoweŚrednie tętno

Wartość uśredniona
danych PM

Skurczowe
Rozkurczowe
TętnoNajnowsze dane

Ostatnie dane (Nr. 1)
Dane pomiarowe zostaną wyświetlone trzy
sekundy po wyświetleniu numeru danych.
3. Po wyświetleniu ostatnich danych wcisnąć
przycisk , aby wrócić do wyświetlaniauśrednionej wartości wszystkich pomiarów.
4. Nacisnąć przycisk START , aby wyłączyć
urządzenie. Po minucie bezczynności
urządzenie wyłączy się automatycznie.
Polski 18

Skurczowe
RozkurczoweTętno

Najstarsze daneCzym jest wskaźnik IHB/AFib?
Gdy urządzenie wykryje nieregularne bicie serca podczas pomiaru, nawyświetlaczu pojawi się wskaźnik IHB/AFib wraz ze zmierzonymi wartościami.
Uwaga: jeśli wskaźnik IHB/AFibbędzie pojawiał się często, zalecamyskonsultowanie się z lekarzem.

Czym jest migotanie
przedsionków (AFib)?
Serce kurczy się pod wpływem bodźców elektrycznych powstających
w komórkach serca, pompując krew. Migotanie przedsionków (Arterial fibrillation,AFib) występuje, gdy bodziec elektryczny w przedsionku ulega zaburzeniom, coz kolei prowadzi do nieregularności pracy serca. Migotanie przedsionków (AFib)może być przyczyną wstrzymania obiegu krwi w sercu, co może doprowadzić do
powstania zakrzepów, a w konsekwencji do udaru lub zawału serca.

Wskaźnik słupkowy ciśnienia
Ten wskaźnik monitoruje zmiany ciśnienia podczas pomiaru.
Nadmuchiwanie

Spuszczanie powietrza

Nadmuchiwanie w toku

Nadmuchiwaniezakończone

Pomiar w tokuWskaźnik klasyfikacji WHOKażdy segment wskaźnika słupkowego
odpowiada klasyfikacji ciśnienia tętniczego
WHO opisanej na kolejnej stronie.
Wskaźnik klasyfikacji WHO
Nadciśnienie ostre
Nadciśnienie umiarkowane
Nadciśnienie łagodne
Prawidłowe wysokie ciśnienie
Normalne ciśnienie
Optymalne ciśnienie

Przykład:

Nadciśnienie umiarkowane

Nadciśnienie łagodne: Na podstawie bieżących danych wskaźnik
wyświetla segment odpowiadający
klasyfikacji WHO.
Polski 19

Prawidłowe wysokie ciśnienie

Informacje o ciśnieniu tętniczym
Czym jest ciśnienie tętnicze?

Ciśnienie tętnicze jest siłą wywieraną przez krew na ścianki tętnic. Ciśnienie
skurczowe występuje podczas kurczenia się przedsionków serca. Ciśnienie
rozkurczowe występuje natomiast podczas rozkurczania się przedsionków.
Ciśnienie tętnicze wyraża się w milimetrach słupa rtęci (mmHg). Naturalneciśnienie tętnicze jest przedstawiane jako ciśnienie bazowe, mierzone
bezpośrednio po obudzeniu się, w stanie spoczynku, przed posiłkiem.

Czym jest nadciśnienie i jak można je kontrolować?
Nadciśnienie jest nienormalnie wysokim ciśnieniem tętniczym, które, w
przypadku braku kontroli, może prowadzić do wielu problemów zdrowotnych,
takich jak udar lub zawał serca. Nadciśnienie można kontrolować, zmieniając
nawyki, unikając stresu oraz przyjmując leki pod kontrolą lekarza.
Stosowanie się do poniższych zaleceń pozwoli kontrolować nadciśnienie lub
zapobiec jego wystąpieniu:
Nie palić tytoniu
Regularnie ćwiczyć
Zmniejszyć spożycie soli i tłuszczu
Poddawać się regularnym
Dbać o prawidłową masę ciałabadaniom lekarskim

Dlaczego warto mierzyć ciśnienie tętnicze w warunkachdomowych?
Pomiar ciśnienia tętniczego w przychodni lub gabinecie lekarskim możewskazywać podwyższone wartości, które mogą być o 25–30 mmHg wyższe niż
wskazane podczas pomiaru w warunkach domowych. Pomiar w warunkachdomowych może ograniczyć wpływ czynników zewnętrznych na ciśnienietętnicze, a ponadto może uzupełniać pomiary dokonywane w gabinecie
lekarskim i pozwolić na uzyskanie dokładniejszej i kompletnej historii pomiarów.

Klasyfikacja ciśnienia tętniczego wg WHO

Zmiany ciśnieniatętniczego

Rozkurczowe ciśnienie tętniczeŚwiatowa
Organizacja Materiał referencyjny: Journal of Hypertension 1999, tom 17 nr 2
Nadciśnienie 3. poziomu (ostre)
Zdrowia(WorldHealth mmHg
110
Organization, WHO) ustaliła
Nadciśnienie 2. poziomu (umiarkowane)
105
normy
oceny
ciśnienia
100tętniczego — przedstawiono je
95 Nadciśnienie 1. poziomu (łagodne)
na wykresie.
90 Prawidłowe wysokie ciśnienie
85
Normalne ciśnienie
80 OptymalneciśnienieRóżnice poziomu ciśnienia
120 130 140 150 160 170 180tętniczego mogą być znaczne
Skurczowe ciśnienie tętnicze w mmHg
— zarówno w ciągu dnia, jaki dłuższego okresu. Różnica dzienna może wynosić od 30 do 50 mmHg.
U osób z nadciśnieniem różnice te są jeszcze wyraźniejsze. Zwykle ciśnienietętnicze wzrasta podczas pracy lub aktywności i spada do najniższego
poziomu podczas snu. Wynik jednego pomiaru nie powinien być zatem
przyczyną obaw.

Polski 20

Typowe wahania w ciągu dnia
(pomiar co pięć minut)Ciśnienie tętnicze

Pomiarów należy dokonywać o tej samej porzednia, postępując zgodnie ze wskazówkami
zamieszczonymi w niniejszej instrukcji. Pozwoli tookreślić normalny poziom ciśnienia tętniczego.
Regularne pomiary pozwolą uzyskać dokładną
historię wartości ciśnienia tętniczego. Za każdym
razem należy odnotowywać godzinę i datępomiaru ciśnienia tętniczego. Aby prawidłowo
zinterpretować
dane
dotyczące
ciśnieniatętniczego, należy skonsultować się z lekarzem.

SkurczoweSen

Rozkurczowe

PMAM

GodzinaRozwiązywanie problemów
ProblemEkran pozostaje
pusty mimowłączeniaurządzenia.

Prawdopodobna
przyczyna
Baterie są rozładowane.
Bieguny baterii nie są
ustawione prawidłowo.

Zalecane działanie
Wymienić wszystkie baterie na
nowe.
Włożyć ponownie baterie
do komory, ustawiając ich bieguny
zgodnie z informacjami w komorze.

Zbyt niskie napięcie baterii.
(NISKI
SymbolMankiet nie
POZIOM BATERII) miga.
Wymienić wszystkie baterie na
napompowuje
W przypadku całkowitego
nowe.
się.
rozładowania bateriiwskaźnik ten nie pojawi się.
Mankiet nie zostałZapiąć mankiet prawidłowo.
prawidłowo zapięty.
Podczas pomiaru należy
Poruszenie ręką lub
pozostawać w bezruchu i nie
ciałem podczas pomiaru.
Urządzenie nie
rozmawiać.
wykonujeUsiąść wygodnie i w bezruchu.
pomiaru.
Pozycja mankietu jest
Oprzeć rękę na stole z dłonią
Odczyty są zbyt
nieprawidłowa.
skierowaną ku górze i mankietem
wysokie lub zbytzałożonym na wysokości serca.
niskie.
Jeżeli tętno jest bardzo słabe lub
nieregularne, urządzenie możemieć problemy z określeniemciśnienia tętniczego.
Wartość różni się od
Zapoznać się z punktem „Dlaczegozmierzonej w przychodni
warto mierzyć ciśnienie tętnicze w
lub w gabinecie lekarskim. warunkach domowych?”.
PozostałeWyjąć baterie. Następnie włożyć
je prawidłowo i ponowić pomiar.
Uwaga: jeśli czynności opisane powyżej nie rozwiążą problemu, należy
skontaktować się ze sprzedawcą. Nie otwierać ani naprawiać produktu
samodzielnie, gdyż spowoduje to unieważnienie gwarancji.
Polski 21

Konserwacja
Nie należy otwierać urządzenia. Zawiera ono delikatne podzespoły elektryczne
oraz skomplikowany system powietrzny, które mogą łatwo ulec uszkodzeniu.
Jeżeli rozwiązanie problemu na podstawie wskazówek dotyczących
rozwiązywania problemów nie powiedzie się, należy skontaktować sięz autoryzowanym dystrybutorem lub działem obsługi klienta. Dział obsługi
klienta firmy A&D udzieli informacji technicznych, informacji o częściachzamiennych oraz urządzeniach autoryzowanym dystrybutorom.
Urządzenie zostało zaprojektowane i wyprodukowane tak, aby zapewnić jak
najdłuższy czas eksploatacji. Mimo to, aby zapewnić prawidłowe
funkcjonowanie i dokładność pomiarów, zaleca się przeprowadzenie kontroliciśnieniomierza co 2 lata. W tym celu należy skontaktować się z najbliższym
autoryzowanym dystrybutorem firmy A&D.

Dane techniczne
Typ
Metoda pomiaru
Zakres pomiaru

Dokładność pomiaru
Zasilanie
Liczba pomiarów

Klasyfikacja

Test kliniczny

EMDPamięć
Warunki pracy
Warunki transportu/
magazynowania
Wymiary

UA-1020-W
pomiar oscylometrycznyCiśnienie: 0–299 mmHg
Ciśnienie skurczowe: 60–279 mmHg
Ciśnienie rozkurczowe: 40–200 mmHg
Tętno:
40–180 ud./minciśnienie: ±3 mmHg
Tętno:±5%
4 baterie 1,5 V (R6P, LR6 lub AA) lub
Zasilacz sieciowy (TB-233C) (niedołączony dozestawu)Ok. 1000 LR6 (baterie alkaliczne)
Ok. 300
R6P (baterie manganowe)
Przy ciśnieniu wynoszącym 180 mmHg i temperaturze
pokojowej 23°C.
elektryczne urządzenie medyczne zasilanewewnętrznie (zasilanie bateriami)/klasa II (zasilacz sieciowy)
Stały tryb pracy
wg ISO81060-2：2013
Podczas klinicznego badania walidacyjnego, model K5
był stosowany u 85 uczestników w celu określenia
rozkurczowego ciśnienia tętniczego.
IEC 60601-1-2: 2014
90 pomiarów
Od +10 do +40°C/od 15 do 85% wilgotności wzgl./od
800 do 1060 hPa
Od -20 do +60°C/od 10 do 95% wilgotności wzgl./od
700 do 1060 hPa
Ok. 140 [szer.] x 60 [wys.] x 105 [gł.] mm
Polski 22

Masa
Zastosowana część
Ochrona
Okres użytkowania

Ok. 285 g bez baterii
mankiet typ BF
urządzenie: IP20urządzenie: 5 lat (w przypadku używania sześć razydziennie)
Mankiet: 2 lata (w przypadku używania sześć razydziennie)
Zasilacz sieciowy: 5 lat (w przypadku używania sześć
razy dziennie)

Dodatkowy zasilacz sieciowy Zasilacz służy do podłączenia monitora ciśnieniatętniczego do domowego gniazda sieciowego.
TB-233C
W celu zakupu prosimy skontaktować się z lokalnym
dystrybutorem firmy A&D.
Zasilacz należy poddawać okresowej kontroli lub
wymianie.
Symbole na zasilaczu sieciowym
SymboleFunkcja/znaczenieWyłącznie do zastosowania wewnątrz
pomieszczeń
Urządzenie klasy II
Bezpiecznik termiczny
Bezpiecznik
Etykieta urządzenia z certyfikatem
zgodności z dyrektywą WE
Etykieta urządzenia z certyfikatem
zgodności EAC
Polaryzacja zasilacza sieciowego
Akcesoria sprzedawane oddzielnie
MankietNumer
katalogowy
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

RozmiarmankietuDuży mankiet dla
dorosłych
Mankieto zwiększonym
zakresie
Mankiet dla
dorosłych
Mały mankiet dla
dorosłych

Obwód ramienia
Od 31 cm do 45 cm
Od 22 cm do 42 cm
Od 22 cm do 32 cm
Od 16 cm do 24 cm

Zasilacz sieciowy
Numer katalogowy
TB-233C

Wtyczka
Typ C

Uwaga: specyfikacja może ulec zmianie bez wcześniejszego powiadomienia.
Klasa IP odpowiada stopniowi ochrony obudowy wg normy IEC 60529. Tourządzenie jest chronione przed ciałami stałymi o średnicy 12 mm i większej(np. palcami). Urządzenie niezabezpieczone przez działaniem wody.
Polski 23

NOTATKI

Polski 24

目錄

致尊敬的顧客 .................................................................................... 2
前言.......................................................................................... 2
用前須知 ........................................................................................... 2部件說明 ........................................................................................... 4
標識含意 ........................................................................................... 5
操作方法 ........................................................................................... 7
使用血壓計 ........................................................................................ 8
安裝／更換電池 ........................................................................... 8接充氣管 ...................................................................................... 9連接交流電轉接器 ....................................................................... 9調整內置的時鐘 ......................................................................... 10選擇正確的袖帶尺寸 .................................................................. 11繞緊袖帶 .................................................................................... 11
怎樣正確測量 ............................................................................. 13
測量 ........................................................................................... 13
測量結束後 ................................................................................ 13
測量方法 ......................................................................................... 14
怎樣正確測量血壓 ..................................................................... 14
以設定的壓力測量您的血壓 ....................................................... 15
以合理的收縮壓測量您的血壓 ................................................... 15
注意正確的測量方法 .................................................................. 16
TriCheckTM 測量 ............................................................................. 16查詢儲存數據 .................................................................................. 18
什麼是 IHB/AFib 指示器？ ............................................................. 19
什麼是 AFib？ ................................................................................ 19
血壓柱指示器 .................................................................................. 19
WHO 分類指示器 ............................................................................ 19
有關血壓的一些知識 ....................................................................... 20
什麼是血壓？ ............................................................................. 20
什麼是高血壓？如何控制高血壓？ ............................................ 20
為什麼在家裡測量血壓？ ........................................................... 20
WHO(世界衛生組織)血壓分類標準 ............................................ 20
血壓差異 .................................................................................... 20
故障檢修 ......................................................................................... 21
維修和保養 ...................................................................................... 22
技術資料 ......................................................................................... 22

中文 1

致尊敬的顧客
銘謝惠顧最新型的A&D血壓計。
薦正確和方便使用本血壓計、並使之成薦您每天血壓管理的最佳助手、敬請在初
次使用前、詳細閱讀此說明書。

前言本血壓計符合歐盟管理文件93/43EEC的有關醫療產品之相關規定﹐且有關標記為證。(0123：有關通知本身的參照號碼。)
權威機構認定的本血壓計適用於成年人，而非新生兒或嬰兒。
應用環境﹕本血壓計適用於在家庭保健環境下自行操作。
本血壓計專用於測量血壓與脈搏次數，供診斷用。

用前須知
本血壓計由精密部件製造。 盡量避免過度曝露於下述環境中，諸如高溫，高濕，
陽光直射，震盪或灰塵。
用乾軟布，或是蘸水和中性洗滌劑的布清潔裝置及袖帶。切勿使用酒精、苯、
稀釋劑或其他刺激性的化學製品來清潔裝置或袖帶。
避免過度用力折疊袖帶或儲存過度扭曲的軟管，放置很長時間。因為這樣會縮短相關設備的壽命。
小心避免軟管與連接線意外勒住嬰幼兒。
測量過程中請勿扭擰充氣管，否則可能因為持續的袖帶壓力造成傷害。
本血壓計和袖帶不耐水，因此應避免接觸水源。
假若血壓計使用時靠近電視、微波爐、行動電話、Ｘ光或其他強電場時，其測量將會模糊不準確。
無線通訊裝置，例如家用網路裝置、行動電話、無線電話及其基座主機，以及對講機，可能會影響本血壓計。因此，本產品應該與這些裝置保持至少 30 cm 的距離。
用過的設備，零部件和電池不能作為一般家庭廢物處理，而應當按照相關的規
定進行處理。
使用交流電轉接器時，請確認交流電轉接器可在必要時隨時從插座拔下。
再次使用血壓計時，請確認血壓計是乾淨的。
請勿改造血壓計，否則可能造成意外，或損壞血壓計。
若要測量血壓，袖帶必須要適當束緊手臂，暫停血液通過動脈。這可能會造成
疼痛、麻木或手臂出現暫時性紅印。此狀況特別會在連續測量時出現。
任何疼痛、麻木或暫時性紅印將會隨時間消失。
測量血壓過於頻繁可能因為血流干擾造成危害。重複使用裝置時，請檢查裝置的運作不會導致血液循環的長期損害。
中文 2

如果您動過乳房切除手術，請在使用前諮詢醫生。
請勿讓兒童自己使用裝置，也請不要在幼兒接觸得到的地方使用裝置。這可能
造成事故或傷害。
內含小配件，如果幼兒誤食，可能會導致窒息。
測量期間未使用時請拔掉交流電適配器。
使用本手冊中未詳細說明的配件可能危及安全性。
若電池發生短路，可能變燙且可能造成燙傷。
使用前（約一小時）請讓裝置適應周圍環境。
尚未對新生兒及孕婦進行臨床試驗。請勿用於新生兒或孕婦。
請勿同時觸摸電池、直流插孔及患者。這可能導致觸電。
請勿在未包裹上臂袖帶的狀況下充氣。

禁忌
以下是正確使用血壓計的注意事項。
請勿將袖帶套到還縛上其他醫療電氣設備的的手臂，否則設備可能無法正常運作。
手臂血液循環不良的病患，請務必在使用血壓計前諮詢醫生，避免醫療糾紛。
請勿自我診斷測量結果，並開始自我治療。請務必向醫生諮詢結果評估與治療。
請勿將袖帶套到有傷口的手臂上。
請勿將袖帶套到接受靜脈滴注或輸血的手臂，否則可能造成傷害或意外。
請勿在有易燃氣體的地方使用血壓計，例如麻醉氣體，否則可能會導致爆炸。
請勿在高度濃縮的氧氣環境中使用血壓計，例如一個高壓氧氣室或氧帳，否則
可能引起火災或爆炸。

中文 3

部件說明
直流插座

顯示屏

按鈕
按鈕
設定 按鈕

充氣管連接插頭
START (開始)按鈕
充氣管插座
充氣管
適當的穿套範圍動脈位置標記
袖

帶

標誌標記
電 池 蓋
電 池 盒

1.5V電 池(R6P，LR6或AA)顯示器的名稱

存儲器

平均

壓力設定

IHB/AFib 標識
袖帶調整錯誤標識

收 縮 壓

移動錯誤標識TriCheckTM標識

舒 張 壓

心臟標識
WHO分類指示器 和
血壓柱指示器
電池指示器

脈 搏 率

AM / PM 標識

時鐘顯示
中文 4

標識含意
印在血壓計器與交流電轉接器的標識
功能 / 意義

標識

建議動作

開關
裝填電池的指示方向
直流電源
SN

系列號碼
生產日期
BF型: 血壓計， 袖帶和相應的管子 已被設計對
電震動有特殊保護作用。
EC管理機構認可的醫療用儀器標識
WEEE標識生產商
EU代表
參閱說明手冊直流插座極性IP

國際保護標識保持乾燥顯示在顯示器上的標識
標識

功能 / 意義

建議動作

當進行測量時即顯示本圖像。當測到脈搏時本圖 繼續測量，保持
像即開始閃爍。
安靜。
IHB/AFib 標識
偵測到心律不整時，即進行顯示。偵測到輕微震動如顫抖或顫動時，燈即亮起。
TriCheckTM 模式自動進行三次連續測量，並顯示
三次測量的平均值。
可能會讀取出錯誤值。
偵測到身體或手臂移動時，即顯示該標識。
再次測量並且在測量時保持安靜。
可能會讀取出錯誤值。
測量時若袖帶未繞緊，即顯示該標識。
正確拉緊袖帶
並再次測量。
中文 5

標識含意
功能 / 意義

標識

建議動作

前次測量的結果儲存於此平均數據
電池電量充足
在測量時電池電壓顯示器即顯示本圖像。
電池電量低
閃爍時更換新電池。
電池電壓太低時閃爍，即顯示本圖像。
在測量過程中由於身體移動導致血壓值不穩定時出現 再次測量並且在本標記。
測量時保持安靜。
當測量收縮壓和舒張壓時，都在10mmHg內時顯示本圖像。
袖帶加壓時，血壓值不增加時顯示本標記。
當袖帶不能正確拉緊時

正確拉緊袖帶並
再次測量。

脈搏顯示錯誤當脈搏不能被正確測到時

請移除電池並按下 START(開始) 按鈕，然後
再次安裝電池。
如仍出現錯誤，請聯繫經銷商。

血壓計內部錯誤SYS收縮壓以mmHg表示

DIA

舒張壓以mmHg表示

PUL脈搏每分鐘速率AM

取自的數據介於 4:00 與 9:59 之間

PM

取自的數據介於 18:00 與 1:59 之間
壓力設定顯示使用者先前設定的壓力值。
中文 6

操作方法
1. 平時測量的保存數據
請按下 START(開始)按鈕。開始測量， 同時血壓計開始記錄保存內容。本血壓計的各按鈕能保存最近的數據90組。

2. 查詢數據
請按下
或按鈕叫出數據。
如圖右所示， 所有測量數據的平均值將顯示在畫面上。

接著， 每次按下顯示。

按鈕時， 儲存數據會如下

取自的所有 AM (上午)測量平均值介於 4:00
與 9:59 之間。
取自的所有 PM (傍晚)測量平均值介於
18:00 與 1:59 之間。
最新數據 (No.n，例如：No.35)
最後一筆數據 (No.1)
欲知如何查詢數據的詳情，請參考“查詢儲存數據”。 (請參照第18頁。)

3. 刪除所有儲存數據同時按下與按鈕，即顯示標識和電池指示器。同時按住和按鈕，直到發亮
的
標識開始閃爍，即可刪除所有儲存數據。

4. 用預設的收縮壓測量
有關預設的收縮壓進行測量時，請參照第15頁。

中文 7

使用血壓計安裝／更換電池
步驟11. 移除電池蓋。
步驟32. 取出舊電池，裝入新的電池時， 要留意
電極(＋，－) 插入新電池（如圖所示）。
僅限於使用R6P，LR6或AA電池。

步驟1

步驟2

3. 闔上電池蓋。閃爍

注意
插入電池時，要按照電池盒内的電極方向，否則血壓計不能工作。
當 （低電壓標記）在顯示屏上閃爍時，同時要用新電池更換全部舊電
池。不要新舊電池一起混用。
血壓計關閉 2 秒後，請更換電池。
更換電池後，如果（低電壓標記）仍顯示，請進行一次血壓測量。
稍後血壓計會進行新電池辨識。
當電池用盡時，（低電壓標記）並不會出現。

電池壽命與環境溫度有關，在低溫情況下，壽命縮短。如果如果每日測量兩次，
一般情況下，四顆新的 LR6 電池電力可持續約一年。
請使用指定的電池。與血壓計同時供應的電池是測試用的,壽命有限。
當血壓計長時間不用時應取出電池。電池液漏出會造成故障。

中文 8

使用血壓計接充氣管將充氣管連接插頭緊緊插入充氣管
插座內。

充氣管插座

充氣管連接插頭

連接交流電轉接器
將交流電轉接器插頭插入直流插座。
之後將交流電轉接器插入電源插座。

交流電轉接器插頭

請使用指定的交流電轉接器。
(請參照第23頁。)

直流插座

若要從電源插座上拔下交流電變壓器，請抓住交流電變壓器機身並從插座拔下。
若要從血壓計拔下交流電變壓器插頭，請抓住交流電變壓器插頭並從血壓計拔下。

中文 9

使用血壓計調整內置的時鐘
使用前 請先進行調整。

按下
或

1. 按下 設定 按鈕直到年份出現。

年或按鈕選擇年份。
2. 用
按下 設定 按鈕設定現在的年份，再選擇月／日。
選擇期間為2010~2059年。

或
月

3. 用
或
按鈕選擇月份。
按下 設定 按鈕設定現在的月份，再選擇日期。
或或按鈕選擇日期。
4. 用
按下 設定 按鈕設定現在的日期，再選擇時／分。
或按鈕選擇小時。
5. 用
按下 設定 按鈕設定現在的小時，再選擇分鐘。
或按鈕選擇分鐘。
6. 用(按下 設定 按鈕，進入壓力設定模式。相關詳盡資訊，請參閱第15頁。)
按下 START(開始) 按鈕即可關閉血壓計。

日

或

時

分

或

註: 超過3分鐘未進行操作，即自動關閉血壓計。
時鐘尚未設定時，時鐘顯示會顯
示如右側顯示的破折號。
按住或按鈕，將連續變更該值。

中文 10

關閉
按下 START(開始)按鈕，即可隨時關閉血壓計。

壓力設定模式使用血壓計選擇正確的袖帶尺寸使用正確的袖帶尺寸至關重要，這樣才能取得正確的讀數。如果袖帶的尺寸不正
確，讀數可能不正確。
手臂尺寸列印在每個袖帶上。
您可從袖帶上的標誌與適當的穿套範圍判斷是否套上正確的袖帶。(請參閱下
一頁的「印在袖帶上的標識」)
如果
標誌位置超過範圍，請聯絡當地的經銷商，購買替換袖帶。
袖帶是消耗品。如果變得破舊，請購買新品。
手臂尺寸
31 cm 到 45 cm
22 cm 到 42 cm
22 cm 到 32 cm
16 cm 到 24 cm
手臂尺寸: 二頭肌的圓周。

推薦袖帶尺寸成人大號袖帶
寬版袖帶
成人袖帶
成人小號袖帶

目錄編號
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SA

不要過緊捲起袖子。

繞緊袖帶
1. 將袖帶繞緊在上臂，距離手肘內側上方約
1-2 cm，如圖所示。
將袖帶直接置於皮膚上，因為衣物可能導致脈搏微弱，造成測量錯誤。

袖帶充氣管
動脈位置標記2. 襯衫的袖子緊緊向上捲起，導致上臂感覺壓迫，可能無法測出正確讀數。
3. 確認
內。

緊緊固定好
織物搭扣。

標誌位置位於適當的穿套範圍

標誌

中文 11

1-2 cm與適當的穿套範圍

使用血壓計印在袖帶上的標識
標識

功能 / 意義

推薦解決方法將
標記置於上臂的動脈
上，或與手臂內側的無名指對齊。

動脈位置標記
▲
REF

標記
目錄編號
適用於成人袖帶的適當穿套範圍。
印在成人袖帶上。
超出範圍，印在成人袖帶與寬版袖帶上。
適用於寬版袖帶的適當穿套範圍。
印在寬版袖帶上。
適用於成人大號袖帶的適當穿套範圍。
印在成人大號袖帶上。
未達範圍，印在成人袖帶與寬版袖帶上。
適用於成人小號袖帶的適當穿套範圍。
印在成人小號袖帶上。

請改用成人大號袖帶，不要用成人袖帶或寬版袖帶。

請改用成人小號袖帶，不要用成人袖帶或寬版袖帶。
請改用成人袖帶，不要用成人大號袖帶與 成人小 號袖帶。

印在成人大號袖帶與成人小號袖帶上。
批號適當的穿套範圍
成人大號袖帶

A

L

寬版袖帶

S

W

L

成人袖帶

S

A

L

S

A

成人小號袖帶

中文 12

使用血壓計怎樣正確測量為了準確的測量血壓，應注意下列幾點。
放鬆坐好。將手臂放在桌上。請勿翹腳，兩腳平放坐直。
測量前先放鬆5～10分鍾。
將袖帶的中心放置於相當與您的心臟同一高度。
測量時保持安靜。
體育活動或洗澡後不要馬上測量。休息二十或三十分鐘再測量。
儘量在每天的同一時間測量血壓。

測量在測量血壓時，通常要感到袖帶很緊方可(請不要驚慌) 。

測量結束後
測量結束後，按下 START(開始) 按鈕即可關閉血壓計。
取下袖帶並記錄測量數據。超過 1 分鐘未進行操作，即自動關閉血壓計。

中文 13

測量方法
進行測量之前，請參照第16頁“注意正確的測量方法”。

怎樣正確測量血壓
1. 將袖帶放於上臂(最好是左上臂)，測量時保持安靜。
2.

按下 START(開始) 按鈕。
所有的顯示部分便會簡短地顯示。畫面上隨即短
暫閃爍 0。
之後開始測量， 如右圖所示顯示器開始變化。
袖帶開始膨脹。袖帶感到有些緊張這是正常的。
在袖帶膨脹的時候，顯示幕左側將出現血壓柱指示符號。

註: 假若您想隨時停止充氣，只需再次按下 START(開始) 按鈕。
3. 當充氣完成時，放氣自動開始，且(心臟標識)
閃爍，表明測量在進行中。
一旦測到脈搏，心臟標記將隨著脈搏的跳動而閃爍。

與心臟同高度

按下

START

所有的顯示
部分便會簡
短地顯示。

顯示 0
開始膨脹

加 壓

註: 假若充氣不完全或不能得到合適的壓力，血壓計將自動再次充氣。
若要避免再次充氣，請參考下一頁的 “以設定的壓力測量您的血壓” 。
4.

當測量結束後，收縮壓和舒張壓讀數及脈搏率在顯示屏上顯示。
袖帶完全排出剩餘的空氣。

5. 按下 START(開始) 即可關閉血壓計。
超過1分鐘未進行操作，即自動關閉血壓計。
註﹕同一個人兩次測量之間應相隔3分鐘以上。

測量進行中

收 縮 壓
舒 張 壓
WHO分類
脈 搏 率
自動排除剩餘的空氣

中文 14

測量方法
本血壓計UA-1020-W被設計用來測量脈博和使袖帶自動充氣到收縮壓水平。
如果發生重複再次加壓情形，請使用下列方式。

以設定的壓力測量您的血壓
進行血壓測量時，可能會發生再次充氣情形。設定固定的壓力值，即可避免再
次充氣。
1.

進行時鐘調整程序步驟 6 時 (請參照第10頁)， 按下 設定 按鈕，進入壓力設定模式。
目前的設定會閃爍。

2.

請按下
或
按鈕，即可選取約 30 mmHg
壓力值，或自下列選取您希望的收縮壓。
AUTO : 自動加壓 (預設值)180 : 180 mmHg 壓力值 (固定)210 : 210 mmHg 壓力值 (固定)240 : 240 mmHg 壓力值 (固定)

3.

或

關閉

TriCheck
設定模式

請按下 設定 按鈕，即可進入 TriCheck 設定模式。
請按下 設定 或 START(開始) 按鈕即可關閉血壓計。超過 3 分鐘未進行操
作，即自動關閉血壓計。
將使用新壓力值進行下一次血壓測量。

以合理的收縮壓測量您的血壓
即使依照以上程序已將壓力值設定為 240，仍發生再次充氣情形；或即使將壓力降低為 20 mmHg 或更低，仍無法顯示結果，請使用本方式。
與心臟同高度1. 將袖帶放於上臂(最好是左上臂)。
2. 按住 START(開始) 按鈕直到比您的預測的收縮壓高30~40mmHg的數字出現。
3. 當預測的數字出現時，放開 START(開始) 按鈕，測量即自動開始，然後參考前頁繼續測量您的血壓。

按住START(開始)按鈕

START

當預計的收縮壓出現時放開開始按鈕

測量方法參考前頁。

中文 15

注意正確的測量方法
舒適而又安靜地坐下，請將您的手臂置於桌上且手掌朝上，並且袖帶與心臟同等高度。
開始測量前放鬆大約5～10分鐘。假若由於情緒關係過於激動或壓抑，則測量值將比正常血壓高或低，而脈博比正常要快。
一個人的血壓值通常會有些變化，這些取決於您當時在做什麼和當時的飲食，飲酒對您的血壓有強烈而直接的影響。
本血壓計根據您的心臟跳動而測量血壓。假若您的心臟跳動微弱或不規律時，
血壓計很難測到您的血壓。
假若血壓計檢測狀況異常， 它將會停止測量並顯示錯誤符號。具體細節請參考第6頁標識含意的說明。
本血壓計適用於成人， 小孩使用之前請洽詢有關的醫生。無人陪伴時小孩不
應該使用本血壓計。
過高的溫度或濕度或海拔可能會影響自動血壓計的效能。

TriCheck

TM

測量

TriCheckTM 模式自動進行三次連續測量，並顯示三次測量的平均值。

選取 TriCheckTM 模式

在壓力設定模式，

1.

在壓力設定模式， 按下 設定 按鈕 即可
輸入TriCheckTM 設定模式。目前的設定
會閃爍。

按下

2.

或
按鈕，即可ON (開啟)或按下
OFF (關閉) TriCheckTM模式。
ON: Tri-checkTM模式
OFF: 一般血壓測量模式 (預設值)3.

按下 START(開始) 或 設定 按鈕 即可關閉血壓計。超過 3 分鐘未進行操作，
即自動關閉血壓計。

中文 16目前的設定會閃爍
或變更設定

或 START使用 TriCheckTM 模式進行血壓測量1.

按下 START(開始) 按鈕。
所有的顯示部分便會簡短地顯示。畫面
上隨即短暫閃爍 0。
之後即開始第一次血壓測量。

與心臟同高度

START

按下

2.

當測量結束後，收縮壓和舒張壓讀數及脈搏率在顯示屏上顯示。然後，1 分鐘倒數
計時會開始第二次血壓測量。

所有的顯示
部分便會簡
短地顯示。

3.

1分鐘之後，會開始第二次血壓測量。

第一次血壓測量

4.

當測量結束後，即顯示第一次與第二次血壓測量的平均讀數然後，1分鐘倒數計時
會開始第三次血壓測量。

5.

1分鐘之後，會開始第三次血壓測量。

第一次血壓測量
的結果
1分鐘之後

6.

當測量結束後，會顯示三次血壓測量的平均讀數，並儲存數據。

註:進行血壓測量時，會顯示 TriCeckTM 標識。
若要取消血壓測量，請按下 START(開始) 按鈕. 在這種情況下，無法儲存任何數據。
第一次血壓測量時，如顯示袖帶調整錯
誤標識，請取消血壓測量，重新正確繞緊袖帶，並開始新的血壓測量。
血壓測量後，會儲存三次血壓測量的平均數據。
完成第三次血壓測量前，若取消血壓測量，則無法儲存任何數據。

中文 17

第二次血壓測量

1分鐘之後

第一次與第二次血壓
測量的平均讀數
第三次血壓測量

三次血壓測量的平均讀數查詢儲存數據
註: 本血壓計可將前90次的測量數據儲存於記憶內。
1. 請按下或按鈕。

畫面將顯示所有測量數據的平均值和數據數量。(如無任何數據，會顯示“0”。
請按下，
或 START(開始) 按鈕
即可關閉血壓計。)
按鈕時 (或 按2. 接著， 每次按下鈕，即可以相反順序進行數據顯示)，儲存數據會如下顯示。

按下

或
平均收縮壓
平均舒張壓
平均脈搏

所有數據
的平均值

平均收縮壓

取自的所有 AM (上午)測量平均值介
於 4:00 與 9:59 之間。
(例如， 10測量。如無任何數據，會顯
示“--”。)

平均舒張壓
平均脈搏
AM數據的
平均值

取自的所有 PM (傍晚)測量平均值介
於 18:00 與 1:59 之間。
(例如， 9測量。如無任何數據，會顯
示“--”。)
最新數據 (No.n，例如：No.35)數據號顯示後三秒，即顯示測量數據。

平均收縮壓
平均舒張壓
平均脈搏
PM數據的
平均值

收縮壓

最後一筆數據 (No.1)
數據號顯示後三秒，即顯示測量數據。
3. 顯示出最後一筆數據後，按下按鈕即
可返回所有血壓測量的平均值顯示。

舒張壓
脈搏
最新數據

4. 請按下 START(開始) 按鈕即可關閉血壓計。超過1分鐘未進行操作，即自動關
閉血壓計。

收縮壓
舒張壓
脈搏
最後一筆數據（最舊數據）中文 18

什麼是IHB/AFib 指示器？
當該血壓計在測量期間偵測到心律不整時，IHB/AFib 指示器將伴隨測量值出現
在該顯示屏上。
註﹕如果經常看到 IHB/AFib符號，建議您聯絡醫生。。

什麼是 AFib？
心臟因心臟中產生的電訊號而收縮並將血液送至全身。心房顫動（AFib）發生
在心房的電訊號變得混亂並導致脈搏間隔的干擾時。AFib 可導致血液停滯在心臟中，這可能容易產生血液凝塊，導致中風和心臟病發作。

血壓柱指示器本血壓柱指示器在測量中顯示壓力的變化。
加 壓

排除空氣

膨脹中

測量中
膨脹結束

WHO分類指示器
每個血壓柱指示器分別與下一頁說明的WHO血壓
分類一致。

例:

WHO分類指示器
嚴重高血壓
中度高血壓
輕度高血壓
血壓偏高
正常
最合適

中度高血壓

: 指示器以現在的數據為依據。顯示與WHO
分類一致的區段。

輕度高血壓

血壓偏高

中文 19

有關血壓的一些知識什麼是血壓？
所謂血壓是血液對動脈血管壁所施加的壓力。當心臟收縮時產生的壓力為收縮壓，而當心臟舒張的時候產生的壓力為舒張壓。
血壓以毫米汞柱(mmHg)作為測量單位。當一個人處於安靜狀態和飲食前，
測量到的血壓代表一個人的基礎血壓。

什麼是高血壓？如何控制高血壓？
高血壓是一種動脈異常的高壓力狀態。假若未被及時注意的話，會引起許多的健康問題包括休克和心臟病發作。
高血壓可以通過改變生活方式，避免緊張和在醫生的指導下服用一些藥物而
得到控制。
預防高血壓或者通過下列措施而得到控制。
不要吸煙經常體育鍛鍊
減少食鹽和脂肪的攝入
定期體檢
維持適當的體重

為什麼在家裡測量血壓？
在診所或醫生辦公室測量血壓時容易產生憂慮心理，使血壓讀數比在家庭測量
的數據高25～30mmHg。在家測量時能減少血壓測量時的一些外部影響因素，提供給醫生更準確而又完整的血壓記錄數值。

WHO(世界衛生組織)血壓分類標準
WHO(世界衛生組織)設立
的評價高血壓的標準如下圖(不考慮年齡因素)參考資料：高血壓雜誌1999第17卷第2 期
Ⅲ高血壓(重度)舒張壓

Ⅱ高血壓（中度）

血壓差異
一個人的血壓可因日夜和季節而有些變化。一般由
於日夜因素血壓值相差30～50mmHg.。有高血壓的人，
這種差異可能更加明顯。
通常當工作或運動的時候，血壓值會升高。

Ⅰ高血壓（輕度）

最佳

正
常

高
於
正
常

收縮壓

中文 20

而在睡眠的時候會下降到最低水平。
所以不要太介意一次的測量結果。
使用本手冊所說明的方法， 在每天的
同一時間測量血壓， 不但知道您自己的血壓值， 而且多次的測量值可以讓您更全面的掌握血壓記錄資料。
每次測量記錄您的血壓數值時，請弄清楚測量的時間和日期。請醫生
幫助解釋您的血壓數據。

在一天之內的典型動態圖(每5分鐘測量一次)
收縮壓
血
壓

舒張壓

睡眠狀態
時間

下午

上午

故障檢修
問題
當電源打
開時，顯
示屏上沒
有任何顯
示

問題原因

推薦解決方法

電池用盡

更換全部電池

電池方法放置不正確

放置電池時陰陽極方向要和袖帶上標
的方向一致

袖帶不充
氣

電池電壓太低 ， 出現
(標 更換新電池記閃爍)。假若電池用盡本標記不會出現。

本血壓計
不能測
量，讀數
太高或太
低。

袖帶扎得不夠緊

正確扎緊袖帶

當在測量血壓時，您移動了手 在測量時保持安靜臂或身體。
袖帶位置放置不正確

坐好保持安靜，請將您的手臂置於桌
上且手掌朝上，並且袖帶與心臟同等高度。
假若心臟跳動微弱或不規律性，血壓計也很難測到血壓值

其他測到的血壓值和在診所或醫請參見“為什麼在家裡測量血壓？”
生辦公室測量的血壓值不同。
取出電池，再正確放進去重新測量。

註： 上述的處置還不能解決問題時， 請與購買的商店聯絡。擅自打開本機或進行修理時就會失去售後服務的權利， 煩請留意。
中文 21

維修和保養
請勿打開本血壓計，它是採用精密的電子原件和一些複雜的部件製造的。這些原件容易損害。假若您按照故障檢修還不能排除故障時，請向供應商或A&D服務中心咨詢。A&D將向您提供權威的技術信息，備用部件和原件。
本機是為持久耐用而製造的。因此，為使本機能夠發揮最合適的性能和最準確的
精度，請每2年接受定期檢查。維修， 保養請在購買的商店或A&D進行。

技術資料
類

型

﹕UA-1020-W

測量方法

﹕示波測量

測量範圍

﹕壓力 ﹕
0～299 mmHg
收 縮 壓﹕60～279 mmHg
舒 張 壓﹕40～200 mmHg
脈搏 ﹕40～180 次/分

測量準確性

﹕壓力 ﹕
脈搏 ﹕

電力供給

﹕4×1.5V 電池 (R6P， LR6或AA) 或
交流電轉接器 (TB-233C) (未隨附)測量次數

﹕ 約 1000 次 LR6 (鹼性電池)
約 300 次 R6P (錳電池)壓力值 180 mmHg，室溫 23 °C種﹕內部供電ME（醫電）設備(由電池提供) /Class II (由交流電轉接器提供)
連續操作模式類

臨床試驗±3 mmHg±5 %﹕符合ISO81060-2：2013
在臨床驗證研究中，K5 用於 85 位受試者以測量舒張壓。

EMD﹕IEC 60601-1-2:2014

數據儲存器﹕最近90個的測量值

操作條件+10 至 +40 °C / 15 至 85 %RH / 800 至 1060 hPa

運送 / 儲存條件 -20 至 +60 °C / 10 至 95 % RH / 700 至 1060 hPa
尺寸﹕大約140(W)×60(H)×105(D)毫米
中文 22

重

量

﹕大約285克，不包括電池

防水防塵﹕血壓計: IP20

觸身零件

﹕袖帶

使用壽命

﹕血壓計: 5 年 (若一天使用六次)
袖帶: 2 年 (若一天使用六次)

BF型

交流電轉接器: 5 年 (若一天使用六次)
交流電轉接器TB-233C﹕該交流電轉接器用於家庭連接電源，請和當地的銷售商聯繫，
購買該交流電轉接器。
必須定期檢查或更換交流電轉接器。
印製在交流電轉接器上的符號
標識
功能 / 意義
僅供室內使用Class II 裝置
溫度保險絲
保險絲
EC 指令裝置標籤
EAC 認證裝置標籤
交流電轉接器插頭的極性

附件單獨銷售﹕
袖帶
目錄編號
袖帶尺寸
CUF-F-LA 成人大號袖帶
寬版袖帶
CUF-I
成人袖帶
CUF-F-A
CUF-F-SA 成人小號袖帶
交流電轉接器目錄編號
TB-233C

上臂尺寸
31 cm 到 45 cm
22 cm 到 42 cm
22 cm 到 32 cm
16 cm 到 24 cm

插頭C 型

註﹕有關技術規格可隨時更改，不再提前通知。
異物防護等級是IEC 60529提供的外殼防護等級。本裝置可隔絕直徑12 mm
以上的固體外物，例如手指。本裝置不防水。

中文 23

MEMO

中文 24

İçindekilerDeğerli Müşterilerimiz .......................................................... 2Başlangıç Açıklamaları .......................................................... 2Önlemler .............................................................................. 2
Kontrendikasyonlar .......................................................... 3
Parçaların Tanımı .................................................................. 4
Semboller ............................................................................. 5Çalışma Modu ....................................................................... 7
Monitörü Kullanma ............................................................... 8
Pilleri Takma / Değiştirme ............................................... 8
Hava Hortumunu Bağlama ................................................ 9
AC Adaptörünü Bağlama .................................................. 9
Dahili Saati Ayarlama ..................................................... 10Doğru Manşon Boyutunu Seçme ..................................... 11Manşonu Takma ............................................................ 11Doğru Ölçümleri Yapma ................................................. 13Ölçüm ............................................................................ 13Ölçümden Sonra ............................................................ 13Ölçümler............................................................................. 14
Normal Ölçüm ............................................................... 14
Ayarlanan Basınçla Ölçüm .............................................. 15İstenen Sistolik Basınçla Ölçüm ...................................... 15Doğru Ölçümle İlgili Notlar............................................. 16
TriCheckTM Ölçümü ............................................................. 16
Bellek Verilerini Geri Çağırma.............................................. 18
IHB/AFib Göstergesi Nedir? ................................................. 19
AFib Nedir? ......................................................................... 19
Basınç Çubuğu Göstergesi .................................................. 19
WHO Sınıflandırma Göstergesi ............................................ 19
Tansiyon Hakkında ............................................................. 20
Tansiyon Nedir? ............................................................. 20
Hipertansiyon Nedir ve Nasıl Kontrol Edilir? .................... 20
Tansiyon Neden Evde Ölçülmelidir? ................................ 20
WHO Tansiyon Sınıflandırması ........................................ 20
Tansiyon Değişimleri ..................................................... 20
Sorun giderme .................................................................... 21
Bakım ................................................................................. 22
Teknik Veriler ..................................................................... 22

Türkçe 1Değerli Müşterilerimiz
Günümüzdeki en gelişmiş monitörlerden biri olan modern A&D tansiyon
monitörünü satın aldığınız için sizi tebrik ederiz. Kullanım kolaylığı vedoğruluk için tasarlanmış bu monitör, günlük tansiyon yönetiminizi
kolaylaştıracaktır.
Aygıtı ilk kullanımınız öncesinde bu kılavuzu dikkatlice okumanızıöneririz.

Başlangıç Açıklamaları
Bu aygıt, Tıbbi Ürünler için Avrupa Direktifi 93/42 EEC’ye uygundur. Bu,
uyumluluk işareti ile belirtilmiştir.
(0123: İlgili yetkili kuruluşa referans numarası)
Aygıt, yenidoğanlarda veya bebeklerde değil, yetişkinlerde kullanım içintasarlanmıştır.
Kullanım ortamı. Bu aygıtı evde bakım sağlanan bir ortamda kendi başınıza
kullanabilirsiniz.
Bu aygıt, tanı konulacak kişilerin tansiyon ve nabız hızını ölçmek içintasarlanmıştır.

Önlemler
Bu aygıtın üretiminde hassas parçalar kullanılmıştır. Sıcaklık uç değerleri,nem, doğrudan güneş ışığı, darbe ve tozdan kaçınılmalıdır.
Aygıtı ve manşonu kuru, yumuşak bir bezle veya su ve yumuşak
deterjanla nemlendirilmiş bir bezle temizleyin. Aygıtı veya manşonu
temizlemek için kesinlikle alkol, benzen, tiner veya başka sert kimyasallarkullanmayın.
Manşonu sıkıca katlamaktan veya hortumu uzun süre sıkı bir şekilde
bükülü tutmaktan kaçının çünkü bu tür davranışlar parçaların ömrünü
kısaltabilir.
Hortumun ve kablonun kazayla bebekleri veya küçük çocuklarıboğmamasına dikkat edin.
Ölçüm sırasında hava hortumunu bükmeyin. Bu durum, sürekli manşon
basıncı nedeniyle yaralanmaya neden olabilir.
Aygıt ve manşon suya dayanıklı değildir. Yağmur, ter ve suyun aygıtı vemanşonu kirletmesine izin vermeyin.
Aygıt televizyonlara, mikrodalga fırınlara, cep telefonlarına, X-ışını veya
güçlü elektrik alanına sahip diğer aygıtlara yakın kullanılırsa ölçümler
bozulabilir.
Ev ağ aygıtları, cep telefonları, kablosuz telefonlar ve baz istasyonları,
telsizler gibi kablosuz iletişim aygıtları bu tansiyon monitörünü etkileyebilir.
Bu yüzden, bu tür aygıtlarla arada minimum 30 cm’lik bir mesafe
korunmalıdır.
Türkçe 2

Kullanılmış ekipman, parçalar ve piller normal ev atıkları olarakdeğerlendirilmez ve uygun yerel düzenlemelere göre atılmalıdır.
AC adaptörünü kullanılırken, AC adaptörünün gerektiğinde hemenelektrik prizinden çıkarılabileceğinden emin olun.
Aygıt yeniden kullanılırken aygıtın temiz olduğunu doğrulayın.
Aygıtta değişiklik yapmayın. Kazalara veya aygıtın hasar görmesine
neden olabilir.
Tansiyonu ölçmek için kol, manşon tarafından arterden kan akışını geçici
olarak durdurmaya yetecek kadar sert sıkılmalıdır. Bu durum kolda acıya,uyuşmaya veya geçici kırmızı izlere neden olabilir. Bu durum özellikleölçüm art arda tekrarlandığında görünür.
Acı, uyuşma veya kırmızı izler zamanla kaybolur.
Tansiyonu çok sık ölçmek kan akışını etkilediğinden zarara neden olabilir.
Aygıtı tekrar tekrar kullanırken aygıtın çalışmasının kan dolaşımını uzunsüre bozmadığını kontrol edin.
Mastektomi olduysanız aygıtı kullanmadan önce lütfen doktora danışın.
Çocukların aygıtı kendi başlarına kullanmasına izin vermeyin ve aygıtı
küçük çocukların erişebileceği yerlerde kullanmayın. Kazalara veya
hasara neden olabilir.
Küçük çocukların yanlışlıkla yutması durumunda boğulma tehlikesine
neden olabilen küçük parçalar içerir.
Ölçüm sırasında kullanılmadığında AC adaptörünü prizden çekin.
Bu kılavuzda belirtilmeyen aksesuarların kullanımı güvenliği tehlikeye
sokabilir.
Pil kısa devre yaparsa ısınabilir ve yanıklara neden olabilir.
Kullanmadan önce aygıtın çevre ortamına adapte olmasına izin verin(yaklaşık bir saat).
Yenidoğan bebeklerde ve hamile kadınlarda klinik test yapılmamıştır.
Yenidoğan bebeklerde veya hamile kadınlarda kullanmayın.
Pillere, DC jakına ve hastaya aynı anda dokunmayın. Bu, elektrikçarpmasıyla sonuçlanabilir.
Manşonu üst kol çevresine sarmadan şişirmeyin.

KontrendikasyonlarAşağıda aygıtın uygun kullanımı ile ilgili önlemler bulunmaktadır.
Manşonu, başka bir elektrikli tıbbi ekipmanın takılı olduğu kola takmayın.
Ekipman düzgün çalışmayabilir.
Kolunda ciddi dolaşım bozukluğu olan hastalar, tıbbi sorunlarlakarşılaşmamak için aygıtı kullanmadan önce bir doktora başvurmalıdır.
Ölçüm sonuçları üzerinden kendi kendinize teşhis koymayın ve tedaviyebaşlamayın. İşlemin sonuçlarının değerlendirilmesi ve tedavi için her
zaman bir doktora başvurun.
İyileşmemiş yarası olan bir kola manşonu takmayın.
İntravenöz damlama veya kan nakli yapılan kola manşonu takmayın.
Yaralanmaya veya kazaya neden olabilir.
Aygıtı anestezik gazlar gibi yanıcı gazların bulunduğu yerlerde
kullanmayın. Patlamaya neden olabilir.
Aygıtı, yüksek basınçlı oksijen odası veya oksijen çadırı gibi fazlakonsantre oksijenli ortamlarda kullanmayın. Yangına veya patlamaya
neden olabilir.
Türkçe 3

Parçaların Tanımı
DC JakıEkranDüğmesiDüğmesiSET Düğmesi

Hava Bağlantı Aparatı

START Düğmesi
Hava Soketi
Hava Hortumu
Uygun Oturtma Aralığı

Arter Konum İşaretiManşonİşaretPil Kapağı
Pil Bölmesi
1,5V Piller(R6P, LR6 veya AA)

Ekran

BELLEK

Ortalama

Basınç Ayarları

IHB/AFib sembolüManşon Oturma Hatası Sembolü

Sistolik
Basınç

Hareket Hatası Sembolü
TriCheckTM Sembolü

Diyastolik
Basınç

Kalp İşareti
WHO Sınıflandırma Göstergesi
ve Basınç Çubuğu Göstergesi
Pil Göstergesi

Nabız Hızı

AM / PM İşareti
Türkçe 4

Saat Ekranı

Semboller
Aygıt kasası ve AC adaptörü üzerine basılmış semboller
Semboller

Fonksiyon / Anlamı
Aygıtı bekleme durumuna alma veaygıtın gücünü açma.
Pil takma kılavuzuÖnerilen EylemDoğru akım
SN

Seri numarasıÜretim tarihi
Tip BF: Aygıt, manşon ve boru,
elektrik çarpmalarına karşı özelkoruma sağlayacak şekildetasarlanmıştır.
EC yönergesi tıbbi aygıt etiketi
WEEE etiketi
Üretici
AB temsilcisi
DC jakı polaritesi

IP

Uluslararası koruma sembolü
Kuru tutun

Ekranda görünen semboller
Semboller

Fonksiyon / AnlamıÖlçüm devam ederken görünür.
Nabız algılandığında yanıp söner.

Önerilen EylemÖlçüm sürüyor. Mümkün
olduğunca hareketsiz
kalın.

IHB/AFib sembolü
Düzensiz kalp atışı algılandığında
görünür.
Ürperme veya titreme gibi çok hafif
titreşim algılandığında yanabilir.
TriCheckTM modu
Otomatik olarak art arda üç ölçüm
yapar ve bu üç ölçümün ortalamadeğerlerini görüntüler.
Bir vücut veya kol hareketi
algılandığında görünür.
Türkçe 5

Yanlış değer gösterilebilir.
Başka bir ölçüm alın.
Ölçüm sırasında
hareketsiz kalın.

Semboller

Fonksiyon / AnlamıManşon gevşek takıldığında ölçüm
yapılırken görünür.

Önerilen Eylem
Yanlış değer
gösterilebilir. Manşonu
doğru takın ve başka birölçüm yapın.

BELLEKTE depolanan öncekiölçümler.
Ortalama veri
DOLU PİL
Ölçüm sırasında pil gücü
göstergesi.
DÜŞÜK PİLYanıp söndüğünde pil gücü
azalmıştır.
Ölçüm sırasında hareket nedeniyle
stabil olmayan tansiyon.

İşaret yanıp söndüğünde
tüm pilleri yenileriyledeğiştirin.
Yeniden ölçmeyi
deneyin.
Ölçüm sırasında
hareketsiz kalın.

Sistolik ve diyastolik değerler
arasında en fazla 10 mmHg fark var.
Şişme sırasında basınç değeri
Manşonu doğru takın ve
artmadı.
yeniden ölçmeyi
deneyin.
Manşon düzgün takılmamıştır.
NABIZ GÖRÜNTÜLEME HATASI
Nabız düzgün algılanmamıştır.
Tansiyon monitörü dahili hatası

SYS

mmHg cinsinden sistolik tansiyon

DIA

mmHg cinsinden diyastolik tansiyon

PUL

Dakikadaki nabız

AM

4:00 - 9:59 arasında alınan veriler

PM

18:00 - 1:59 arasında alınan veriler
Basınç ayarları
Daha önce kullanıcı tarafından
ayarlanan basınç değerini gösterir.
Türkçe 6

Pilleri çıkarıp STARTdüğmesine basın ve sonra
pilleri yeniden takın. Hata
gözükmeye devam
ediyorsa bayiye başvurun.

Çalışma Modu
1. Normal Ölçüm
START düğmesine basın. Tansiyon ölçülür ve veri aygıtta depolanır. Bu
aygıt, yapılan son doksan ölçümü bellekte otomatik olarak depolayabilir.

2. Verileri Geri Çağırma
Bellekteki verileri geri çağırmak için
ya dadüğmesine basın. Sağdaki şekilde
gösterildiği gibi tüm ölçümlerin ortalaması
görüntülenir.
Daha sonra,düğmesine her basıldığında
bellek verileri aşağıdaki şekilde görüntülenir.
4:00 - 9:59 arasında yapılan tüm AM(sabah) ölçümlerinin ortalaması.
18:00 - 1:59 arasında yapılan tüm PM(akşam) ölçümlerinin ortalaması.
En yeni veriler (No.n, örnekte No.35)
Son veriler (No.1)
Verileri geri çağırma hakkında ayrıntılar için
“Bellek Verilerini Geri Çağırma” bölümüne
başvurun. (Bkz. sayfa 18.)

3. Bellekte Depolanan Tüm Verileri SilmeHem
hem düğmesine basın.
işareti ve
pil göstergesi görünür. Bellekte depolanan tüm
verileri silmek için, yanan
sönmeye başlayana kadar
aynı anda basılı tutun.

işareti yanıpve düğmelerini

4. İstenen Sistolik Basınçla Ölçüm
İstenen sistolik basınçla ölçüm için sayfa 15’e bakın.
Türkçe 7

Monitörü Kullanma
Pilleri Takma / Değiştirme
1. Pil kapağını çıkarın.
2. Kullanılmış pilleri çıkarın ve yeni
pilleri kutuplarının (+ ve -) doğru
olmasına dikkat ederek gösterildiği
gibi pil bölmesine takın.
Yalnızca R6P, LR6 veya AA pilleri
kullanın.
3. Pil kapağını takın.

Adım 1
Adım 3

Adım 1
Adım 2DİKKAT
Pilleri gösterildiği gibi pil bölmesine takın. Yanlış takılırsa aygıt çalışmaz.
Ekranda(DÜŞÜK PİL işareti) yanıp söndüğünde, tüm pilleri yenileriyledeğiştirin. Eski ve yeni pilleri karıştırmayın. Pil ömrünü azaltabilir veya
aygıtın arızalanmasına neden olabilir.
Pilleri değiştirmek için aygıt kapatıldıktan sonra en az iki saniye bekleyin.
Piller değiştirildikten sonra(DÜŞÜK PİL işareti) görünmeye devam
ediyorsa bir tansiyon ölçümü yapın. Böylece aygıt yeni pilleri tanıyabilir.
Piller bittiğinde
(DÜŞÜK PİL işareti) görünmez.
Pil ömrü ortam sıcaklığına göre değişebilir ve düşük sıcaklıklarda
kısalabilir. Genel olarak, her gün iki kez ölçüm yapıldığında dört adet yeni
LR6 pil yaklaşık bir yıl dayanır.
Yalnızca belirtilen pilleri kullanın. Aygıtla verilen piller monitör
performansını test etmek içindir ve sınırlı ömrü vardır.
Aygıt uzun süre kullanılmayacaksa pilleri çıkarın.
Piller sızıntı yapabilir ve arızaya neden olabilir.

Türkçe 8

Monitörü Kullanma
Hava Hortumunu Bağlama

Hava soketi
r soketi

Hava bağlantı aparatını hava
soketine sıkıca takın.

Hava bağlantı aparatı

AC Adaptörünü Bağlama
AC adaptörü fişini DC jakına takın.
Ardından, AC adaptörünü elektrik
prizine bağlayın.

AC adaptörü fişiBelirtilen AC adaptörünü kullanın.
DC jakı(Bkz. sayfa 23.)
AC adaptörünü elektrik prizinden çıkarırken AC adaptörü gövdesinden
tutup çekerek prizden çıkarın.
AC adaptörü fişini tansiyon monitöründen çıkarırken AC adaptörü fişini
tutup monitörden çıkarın.

Türkçe 9

Monitörü Kullanma
Dahili Saati AyarlamaŞuna basın:

Kullanımdan önce saati ayarlayın.

veya

1. Yıl yanıp sönmeye başlayana kadar SETdüğmesine basın.
2.

3.

Yıl

veya düğmesini kullanarak yılı seçin.
Mevcut yılı ayarlayıp ay/gün seçiminegeçmek için SET düğmesine basın. Tarih,
2010 ve 2059 yılları arasında ayarlanabilir.

veya

Ay

veya düğmesini kullanarak ayı seçin.
Mevcut ayı ayarlayıp gün seçimine geçmek
için SET düğmesine basın.

4.

veya düğmesini kullanarak günü seçin.
Mevcut günü ayarlayıp saat/dakika seçiminegeçmek için SET düğmesine basın.

5.

veya düğmesini kullanarak saati seçin.
Mevcut saati ayarlayıp dakika seçiminegeçmek için SET düğmesine basın.

6.

veya
düğmesini kullanarak dakikayı
seçin.
(Basınç ayar moduna gitmek için SETdüğmesine basın. Ayrıntılar için bkz. sayfa
15.)
Aygıtı kapatmak için START düğmesinebasın.
Üç dakika hiçbir işlem
yapılmazsa aygıt otomatikolarak kapanır.
Saat ayarlanmazsa saattesağda gösterilen çizgiler
görüntülenir.
ya dadüğmesi basılı
tutulursa değer sürekli değişir.

veya

Gün

veya

Saat

Dakikaveya

Not:

Türkçe 10

Kapat
Aygıtı kapatmak
için
istediğiniz
zaman
START
düğmesine basın.

Basınç
ayar modu

Monitörü KullanmaDoğru Manşon Boyutunu SeçmeDoğru manşon boyutunu kullanmak, okunan değerin doğruluğu içinönemlidir. Manşon düzgün boyutta değilse okunan değer yanlış tansiyondeğerini gösterebilir.
Kol boyutu her manşonun üzerine basılmıştır.
Manşon üzerindekiişareti ve uygun oturtma aralığı doğru manşonu
takıp takmadığınızı size belirtir. (Sonraki sayfada “Manşon üzerinde yazılı
semboller” konusuna Tablosuna bakın)işareti aralığın dışını gösteriyorsa yeni bir manşon satın almak için size
yakın bayiye başvurun.
Manşon, bir sarf malzemesidir. Yıpranırsa yenisini satın alın.
Kol BoyutuÖnerilen Manşon Boyutu
31 cm - 45 cm
Büyük Yetişkin Manşonu
22 cm - 42 cm
Geniş Aralıklı Manşon
22 cm - 32 cm
Yetişkin Manşonu
16 cm - 24 cm
Küçük Yetişkin Manşonu
Kol boyutu: Pazı çevresi.

Katalog Numarası
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SAManşonu Takma
1. Manşonu üst kol çevresine, dirsek
içinin yaklaşık 1 - 2 cm üstünden
gösterildiği gibi sarın.
Kıyafetle ölçüm, belirsiz nabza ve
dolayısıyla ölçümün hatayla
sonuçlanmasına neden
olabileceğinden manşonu doğrudan
cildin üzerine yerleştirin.

Giysinizin kolunu çok sıkı kıvırmayın.
Manşon1-2 cmHava Hortumu
Arter konum işareti

2. Giysi
kolunusıkıcayukarı
kıvırmadan
kaynaklananüstkoldaki daralma okunan değerlerindoğruluğunu önleyebilir.
3.

işaretinin uygun oturtma aralığı
içini gösterdiğini onaylayın.

Türkçe 11Kumaş kopçayısağlamca sabitleyin.

işaretive uygun oturtma aralığı

Monitörü KullanmaManşon üzerine basılmış semboller
Semboller Fonksiyon / AnlamıÖnerilen Eylemişaretini üst kolun
arterinde veya kolun içtarafında yüzük parmağıyla
hizalı olarak ayarlayın.

Arter konum işareti▲
REFİşaretKatalog numarasıYetişkin manşonu için uygun oturtma
aralığı.
Yetişkin manşonu üzerine basılıdır.
Yetişkin manşonu veya
Yetişkin manşonu ve geniş aralıklı geniş aralıklı manşon yerinemanşon üzerine basılı aralık üstünde. büyük yetişkin manşonu
kullanın.
Geniş aralıklı manşon için uygun
oturtma aralığı. Geniş aralıklı manşon
üzerine basılıdır.
Büyük yetişkin manşonu için uygun
oturtmaaralığı.
Büyükyetişkinmanşonu üzerine basılıdır.
Yetişkin manşonu veya
Yetişkin manşonu ve geniş aralıklı geniş aralıklı manşon yerinemanşon üzerine basılı aralık altında.
küçük yetişkin manşonu
kullanın.
Küçük yetişkin manşonu için uygun
oturtmaaralığı.
Küçük
yetişkinmanşonu üzerine basılıdır.
Büyük yetişkin manşonu
Büyük yetişkin manşonu ve küçük veya
küçükyetişkinyetişkin manşonu üzerine basılıdır.
manşonu yerine yetişkin
manşonu kullanın.
Parti numarası

Uygun oturtma aralığı
Büyük yetişkin manşonu

A

LGeniş aralıklı manşon

S

W

L

Yetişkin manşonu

S

A

L

S

A

Küçük yetişkin manşonu

Türkçe 12

Monitörü KullanmaDoğru Ölçümleri Yapma
En doğru tansiyon ölçümü için:
Bir sandalye üzerine rahat bir şekilde oturun. Kolunuzu masanın üzerine
koyun. Bacak bacak üzerine atmayın. Ayaklarınızı zeminde düz tutun ve
sırtınızı düzleştirin.
Ölçümden önce beş ila on dakika kadar dinlenin.
Manşonun ortasını kalbinizle aynı düzeye yerleştirin.
Ölçüm sırasında hareketsiz durun ve sessiz kalın.
Fiziki egzersizden veya banyodan yaptıktan hemen sonra ölçüm yapmayın.
Ölçüme başlamadan önce yirmi veya otuz dakika dinlenin.
Tansiyonunuzu her gün aynı saatte ölçmeye çalışın.

ÖlçümÖlçüm sırasında, manşonu çok sıkı olarak hissetmeniz normaldir.
(Paniklemeyin).

Ölçümden SonraÖlçümden sonra aygıtı kapatmak için START düğmesine basın.
Manşonu çıkarın ve verilerinizi kaydedin. Bir dakika hiçbir işlem yapılmazsa
aygıt otomatik olarak kapanır.

Türkçe 13ÖlçümlerÖlçümden önce sayfa 16’daki “Doğru Ölçümle İlgili Notlar” bölümünü
okuyun.

Normal Ölçüm
1.

Manşonu kol üzerine yerleştirin (tercihen sol kol).
Ölçüm sırasında sessizce oturun.

2.

START düğmesine basın.
Tüm ekran segmentleri görüntülenir.
Kısa bir süreliğine yanıp sönen sıfır (0)görüntülenir.
Ölçüm başladığında sağdaki şekilde
gösterildiği gibi ekran değişir. Manşon
şişmeye başlar. Manşonun çok sıktığını
hissetmeniz normaldir. Şişme sırasında
ekranın sol kenarında bir basınç çubuğu
göstergesi görüntülenir.
Not: Şişirmeyi herhangi bir zaman
durdurmak
isterseniz
STARTdüğmesine yeniden basın.

Şuna basın: START
Tüm ekransegmentleri
görüntülenir

Sıfır gösterilirŞişirmeye başlar

3.

Şişme tamamlandığında sönme otomatik(kalp işareti) yanıpolarak başlar vesönerek ölçümün sürdüğünü gösterir.
Nabız algılandığında her nabız atışında
işaret yanıp söner.
Not: Uygun bir basınç sağlanmazsa aygıt
otomatik olarak yeniden şişmeye başlar.
Yeniden şişmeyi önlemek için bir
sonraki sayfada yer alan
“Ayarlanan Basınçla Ölçüm”
konusuna bakın.

4.

Ölçüm tamamlandığında sistolik ve
diyastolik basınç okunan değerleri ve nabız
hızı görüntülenir.
Manşon, kalan havayı boşaltır ve tamamensöner.

5.

Aygıtı kapatmak için START düğmesinebasın.
Bir dakika hiçbir işlem yapılmazsa aygıt
otomatik olarak kapanır.

Not: Aynı kişide ölçüm yaparken ölçümler
arasında en az üç dakika bekleyin.

Türkçe 14

Kalp seviyesinde

Basınç veriliyorÖlçümsürüyor

Sistolik basınç
Diyastolik basınç
WHO sınıflandırması
Nabız hızı
Kalan havayı otomatikolarak boşaltırÖlçümler
UA-1020-W nabzı algılamak ve manşonu sistolik basınç düzeyine
otomatik olarak şişirmek üzere tasarlanmıştır.
Yeniden şişme tekrar tekrar meydana gelirse aşağıdaki yöntemleri kullanın.

Ayarlanan Basınçla Ölçüm
Tansiyon ölçümü sırasında yeniden şişme gerçekleşebilir.
Yeniden şişmeden sakınmak için sabit bir basınç değeri ayarlanabilir.
1.

2.

Saat ayarlama işleminin 6. adımında (bkz.
sayfa 10) basınç ayar moduna gitmek için
SET düğmesine basın. Geçerli ayar yanıp
söner.

veya

Beklenen sistolik basıncınızdan yaklaşık 30
mmHg ve üstü bir basınç değerini aşağıdaki
veya
düğmesineşekilde seçmek içinbasın.
AUTO : Otomatik basınç verme (varsayılan değer)180 : 180 mmHg basınç değeri (sabit)
Kapat
210 : 210 mmHg basınç değeri (sabit)
240 : 240 mmHg basınç değeri (sabit)

3.

TriCheckayar modu

TriCheck ayar moduna gitmek için SET düğmesine basın.
Aygıtı kapatmak için START düğmesine basın. Üç dakika hiçbir işlem
yapılmazsa aygıt otomatik olarak kapanır.
Sonraki ölçüm yeni basınç değeriyle yapılır.

İstenen Sistolik Basınçla Ölçüm
Bu yöntemi, basınç değeri yukarıdaki işlemde 240 olarak ayarlanmasına
veya basıncın 20 mmHg veya daha az bir seviyeye düşmesine rağmen
tekrar tekrar yeniden şişme gerçekleştiğinde kullanın.
1.

Manşonu kol üzerine yerleştirin (tercihen sol kol). Kalp seviyesinde

2.

Beklenensistolikbasıncınızdan
yaklaşık 30 - 40 mmHg yüksek bir
sayı
görünene
kadar
START
düğmesini basılı tutun.

3.

İstediğiniz sayıya erişildiğinde, ölçümübaşlatmak için START düğmesinibırakın. Önceki sayfada açıklandığı gibi
tansiyonunuzu ölçmeye devam edin.
Türkçe 15Düğmeyi basılı tutun STARTİstenen sistolik
basınçta düğmeyi
bırakın
Ölçüm için önceki sayfaya
bakın

Doğru Ölçümle İlgili Notlar

Rahat bir konumda oturun. Avuç içiniz yukarı bakacak şekilde vemanşonu kalbinizle aynı düzeyde tutarak kolunuzu masa üzerine
yerleştirin.
Ölçüme başlamadan önce beş ila on dakika kadar dinlenin.
Heyecanlıysanız veya duygusal baskı altındaysanız bu stres ölçüme
yansıyarak normal tansiyon değerinden daha yüksek (veya daha düşük)bir sonuç çıkmasına neden olabilir ve okunan nabız değeri genellikle
normalden fazla olur.
Bir kişinin tansiyonunu yapılan işe ve yenilenlere bağlı olarak süreklideğişir. İçtiğiniz şeyler, tansiyonunuzu çok güçlü ve hızlı bir şekilde
etkileyebilir.
Bu aygıt, ölçümlerini kalp atışına göre yapar. Çok zayıf veya düzensiz bir
kalp atışınız varsa aygıt, tansiyonunuzu ölçmede zorlanabilir.
Aygıt anormal bir durum algıladığında ölçmeyi durdurur ve bir hatasembolü görüntüler. Sembollerin açıklaması için bkz. sayfa 6.
Bu tansiyon monitörü, yetişkinlerin kullanması için tasarlanmıştır. Bu
aygıtı çocuklar için kullanmadan önce doktorunuza danışın. Çocuklar, bu
aygıtı gözetimsiz kullanmamalıdır.
Otomatik tansiyon monitörünün performansı, aşırı sıcaklık veya nemden
ya da rakımdan etkilenebilir.

TriCheckTM Ölçümü
TriCheckTM modu otomatik olarak art arda üç ölçüm yapar ve bu üçölçümün ortalama değerlerini görüntüler.

TriCheckTM Modunu Seçme

Basınç ayar modunda,TM

1.

Basınç ayar modunda, TriCheckayar moduna gitmek için SETdüğmesine basın. Geçerli ayar yanıp
söner.

2.

TriCheckTM modunu açmak ya daveya
kapatmak için
Geçerli ayar yanıp sönerdüğmesine basın.
AÇIK: TriCheckTM modu
veya
KAPALI: Normal ölçüm modu (varsayılan değer)Ayarı değiştirin
Aygıtı kapatmak için START veya
SET düğmesine basın. Üç dakika
veya START
hiçbir işlem yapılmazsa aygıt
otomatik olarak kapanır.

3.

Türkçe 16şuna basın:

TriCheckTM Modu Kullanarak Ölçüm
1.

START düğmesine basın.

Tüm ekran segmentleri görüntülenir.
Kısa bir süreliğine sıfır (0) yanıp
söner ve ilk ölçüm başlar.
2.

Ölçüm tamamlandığında sistolik ve
diyastolik basınç okunan değerleri venabız hızı görüntülenir ve ardından,
ikinci ölçüm için bir dakikalık geri
sayım başlar.

3.

Bir dakika sonra ikinci ölçüm başlar.

4.

Ölçüm tamamlandığında birinci ve
ikinci ölçümlerin okunan ortalamadeğerleri görüntülenir ve ardından,üçüncü ölçüm için bir dakikalık geri
sayım başlar.

5.

Bir dakika sonra üçüncü ölçüm
başlar.

6.

Ölçüm tamamlandığında üç ölçümün
okunanortalamadeğerleri
görüntülenir ve bellekte depolanır.

Notlar:Ölçüm
sırasında
TriCheckTM
sembolü görüntülenir.
Ölçümü iptal etmek için STARTdüğmesine basın. Bu durumda,
bellekte hiçbir veri depolanmaz.
İlk ölçüm sırasında manşon oturma
hatası sembolü görünürse ölçümü
iptal edin, manşonu yeniden doğru
şekilde takın ve yeni bir ölçüm
başlatın.
Ölçümden sonra üç ölçümün
ortalama verileri bellekte depolanır.
Üçüncü ölçüm tamamlanmadanönce ölçümler iptal edilirken hiçbirveri depolanmaz.

Türkçe 17

Kalp seviyesinde
STARTŞuna basın:

Tüm ekransegmentleri görüntülenirİlk ölçüm

Bir dakika sonraİlk ölçümün
sonuçlarıİkinci ölçüm

Bir dakika sonra

Birinci ve ikinciölçümlerin ortalamasıÜçüncü ölçümÜç ölçümün
ortalaması

Bellek Verilerini Geri Çağırma
Not:
1.

2.

Bu aygıt, yapılan son doksan ölçümü bellekte otomatik olarak depolar.

veya düğmesine basın.
Tüm ölçümlerin ortalaması ve verilerin
sayısı görüntülenir.
(Herhangi bir veri yoksa “0”görüntülenir. Aygıtı kapatmak için ,veya START düğmesine basın.)düğmesine (ya da verileri terssırada görüntülemek içindüğmesine) her basıldığında bellek
verileri aşağıdaki şekilde
görüntülenir.
4:00 - 9:59 arasında yapılan tüm AM(sabah) ölçümlerinin ortalaması.
(Örnekte, 10 ölçüm. Herhangi bir
veri yoksa “--” görüntülenir.)
18:00 - 1:59 arasında yapılan tüm
PM (akşam) ölçümlerinin ortalaması.
(Örnekte, 9 ölçüm. Herhangi bir veri
yoksa “--” görüntülenir.)
En yeni veriler (No.n, örnekte No.35)
Veri numarası ekranından üç saniye
sonra ölçüm verileri görüntülenir.

veyadüğmesine basın.

Ortalama sistolik
Ortalama diyastolik
Ortalama nabız
Tüm verilerin
ortalaması

Ortalama sistolik
Ortalama diyastolik
Ortalama nabız
AM verilerinin
ortalaması
Ortalama sistolik
Ortalama diyastolik
Ortalama nabız
PM verilerinin
ortalaması
Sistolik
Diyastolik

Son veriler (No.1)
Veri numarası ekranından üç saniye
sonra ölçüm verileri görüntülenir.

Nabız

En yeni veriler

3. Son veriler görüntülendikten sonra, tümölçümlerin ortalama ekranına dönmek
içindüğmesine basın.
4. Aygıtı kapatmak için START düğmesinebasın. Bir dakika hiçbir işlem
yapılmazsa aygıt otomatik olarak
kapanır.
Türkçe 18

Sistolik
Diyastolik
Nabız

Son veriler (En eski)

IHB/AFib Göstergesi Nedir?
Monitör, ölçümler sırasında düzensiz ritim algıladığında IHB/AFibgöstergesi, ölçüm değerleriyle birlikte ekranda görüntülenir.
IHB/AFib göstergesini sık sık görüyorsanız doktorunuza
Not: Bubaşvurmanızı öneririz.

AFib Nedir?
Kalp, oluşan elektrik sinyalleri nedeniyle kasılır ve tüm gövdeye kan
gönderir. Atriyumdaki elektrik sinyali karışırsa ve nabız aralıklarında
bozukluğa neden olursa atriyal fibrilasyon (AFib) oluşur. AFib, kanın
kalpte birikmesine neden olarak felç ve kalp krizini tetikleyen kan
pıhtılarına yol açar.

Basınç Çubuğu Göstergesi
Gösterge ölçüm sırasında basıncın ilerleyişini izler.
Basınç veriliyor
Hava salınıyorŞişirme devam ediyorÖlçme devam ediyorŞişirme tamam

WHO Sınıflandırma GöstergesiÇubuk göstergesinin her bir segmenti,
sonraki sayfada tanımlanan WHO tansiyon
sınıflandırmasına karşılık gelir.
WHO Sınıflandırma GöstergesiŞiddetli yüksek tansiyon
Orta derecede yüksek tansiyon
Hafif yüksek tansiyon
Yüksek normal
NormalOptimum: Gösterge geçerli verilere göre WHOsınıflandırmasına karşılık gelen bir
segment görüntüler.

Türkçe 19Örnek:

Orta derecede yüksek tansiyon

Hafif yüksek tansiyon

Yüksek normal

Tansiyon Hakkında
Tansiyon Nedir?
Tansiyon, arter duvarlarına kan tarafından uygulanan kuvvettir. Sistolik
basınç kalp kasıldığında oluşur. Diyastolik basınç kalp genişlediğinde
oluşur. Tansiyon, milimetre cıva cinsinden ölçülür (mmHg). Kişinin doğal
tansiyonu, kişi hâlâ dinlenirken ve yemek yemeden önce sabah ilk işolarak ölçülen temel basınç ile temsil edilir.

Hipertansiyon Nedir ve Nasıl Kontrol Edilir?
Hipertansiyon, anormal şekilde yüksek arter tansiyonudur ve
ilgilenilmezse felç ve kalp krizi dahil birçok sağlık sorununa neden olabilir.
Hipertansiyon, yaşam tarzı değiştirilerek, stresten uzak durularak ve
doktor kontrolünde ilaç tedavisiyle kontrol edilebilir.
Hipertansiyonu önlemek veya kontrol altında tutmak için:
Sigara içmeyin
Düzenli egzersiz yapın
Tuz ve yağ alımınızı azaltın
Düzenli fiziki check-up yaptırın
Uygun ağırlığınızı koruyun

Tansiyon Neden Evde Ölçülmelidir?
Klinikte veya muayenehanede yapılan tansiyon ölçümü endişeye neden
olabilir ve okunan değerler evde ölçülenden 25 - 30 mmHg daha yüksek
çıkabilir. Evde ölçüm, tansiyon okunan değerlerinde dış etkileri azaltır,
doktorun okuduğu değerleri destekler ve daha doğru, tam bir tansiyon
geçmişi sağlar.

WHO Tansiyon Sınıflandırması

Tansiyon Değişimleri

Referans Materyal: Journal of Hypertension 1999, Cilt 17 No.2
mmHg
Derece 3 hipertansiyon (ciddi)
110
Derece 2 hipertansiyon (orta)
105
Diyastolik tansiyon

Dünya Sağlık Örgütü (WHO)tarafından belirlenmiş veçizelgede gösterildiği gibiyaşa bakılmaksızın yüksek
tansiyonudeğerlendiren
standartlardır.

100
95
90
85
80

Derece 1 hipertansiyon (hafif)
Yüksek-normal
NormalOptimum
120 130 140 150 160 170 180
Cinsinden sistolik tansiyon
mmHgKişinin tansiyonu günlük ve
mevsimsel olarak büyükölçüde değişir. Gün boyuncaçeşitli koşullara bağlı olarak 30 ila 50 mmHg arasında değişebilir.
Tansiyonu yüksek kişilerde değişimler daha da fazla olabilir. Tansiyon,
genellikle çalışırken veya oynarken yükselir ve uyku sırasında en düşük
düzeyine düşer. Bu yüzden, tek bir ölçümün sonucu üzerine çok fazlaendişelenmeyin.

Türkçe 20

Gün içinde tipik dalgalanma(Her beş dakikada bir ölçüm)
SistolikTansiyon

Normal tansiyonunuzu öğrenmek için bu
kılavuzda
açıklananprosedürükullanarak her gün aynı saatte ölçüm
yapın. Düzenli yapılan ölçümler dahakapsamlı tansiyon geçmişi sağlar.
Tansiyonunuzu kaydederken tarih ve
saati not ettiğinizden emin olun. Tansiyon
verilerinizi yorumlamak için doktorunuzadanışın.

Uyku

Diyastolik
ÖS

ÖÖ

Saat

Sorun giderme
Sorun
Aygıt açık olsa
bile ekranda
hiçbir şey
görünmüyor.
Manşonşişmiyor.

Aygıt ölçüm
yapmıyor.
Okunandeğerler çok
yüksek veya
çok düşük.

Olası Neden
Piller bitmiştir.
Pil uçları doğru
konumda değildir.
Pil voltajı çok düşüktür.
(DÜŞÜK PİL işareti)
yanıp sönüyor. Piller
tamamen bittiyse işaret
görünmez.
Manşon düzgün şekilde
takılmamıştır.
Ölçüm sırasındakolunuzu veya
gövdenizi hareket
ettirmişsinizdir.

Önerilen Eylem
Tüm pilleri yenileriyle değiştirin.
Pilleri, negatif ve pozitif uçları pilbölmesinde gösterilen şekillerleeşleşecek şekilde yeniden takın.
Tüm pilleri yenileriyle değiştirin.

Manşonu düzgün takın.
Ölçüm sırasında hareketsiz ve
sessiz kaldığınızdan emin olun.

Rahat ve hareketsiz oturun.
Avuç içiniz yukarı bakacakManşon konumu doğru
şekilde ve manşonu kalbinizledeğil.
aynı düzeyde tutarak kolunuzu
masa üzerine yerleştirin.
Çok zayıf veya düzensiz bir kalpatışınız varsa aygıt
tansiyonunuzu ölçmede
zorlanabilir.
Değer, klinikte veya
“Tansiyon Evde Neden
muayenehanedeÖlçülmelidir?” bölümüne
ölçülenden farklı.
başvurun.
Diğer
Pilleri çıkarın. Geri yerine takın
ve başka bir ölçüm yapın.
Not: Yukarıda açıklanan eylemler sorunu çözmezse bayi ile temasa geçin.
Bu ürünün içini açmaya veya onarmaya çalışmayın; bunu yapmaya
kalkarsanız garantiniz geçersiz olur.
Türkçe 21

BakımAygıtın içini açmayın. Hassas elektrikli bileşenler ve hasar görebilecek
karmaşık bir hava ünitesi kullanmaktadır. Sorun giderme talimatlarını
kullanarak sorunu gideremezseniz, bölgenizdeki yetkili bayiye veya
müşteri hizmetleri departmanımıza başvurun. A&D müşteri hizmetleri,yetkili bayilere teknik bilgi, yedek parça ve üniteler sağlar.
Aygıt uzun kullanım ömrü için tasarlanmış ve üretilmiştir. Ancak, aygıtındüzgün ve doğru çalışması için genellikle 2 yılda bir incelenmesi önerilir.
Lütfen bakım için bölgenizdeki yetkili bayiye ya da A&D’ye başvurun.

Teknik Veriler
Tip

UA-1020-WÖlçüm yöntemi

Osilometrik ölçüm

Ölçüm aralığı

Basınç:0 - 299 mmHg
Sistolik basınç: 60 - 279 mmHg
Diyastolik basınç: 40 - 200 mmHg
Nabız:
40 – 180 atış / dakikaÖlçüm doğruluğu

Basınç:Nabız:

Güç kaynağı

4 x 1,5V pil (R6P, LR6 veya AA) veya
AC adaptörü (TB-233C) (Dahil değildir)Ölçüm sayısı

Yakl. 1000 defa
LR6 (alkalin piller)
Yaklaşık 300 defa R6P (manganez piller)
23 °C oda sıcaklığında 180 mmHg basınç ile

Sınıflandırma

Dahili güçle çalışan ME ekipmanı (Pillerle
beslenmiştir) /Sınıf II (Adaptörle beslenmiştir)Sürekli çalışma modu

Klinik test

ISO81060-2：2013’e göre
Klinik doğrulama çalışmasında, diyastolik
tansiyonun tespit edilmesi için 85 gönüllü üzerinde
K5 kullanılmıştır

EMD

IEC 60601-1-2: 2014

BellekSon 90 ölçüm

±3 mmHg%±5

Kullanım koşulları+10 ilâ +40 °C / %15 ilâ 85 BN / 800 ilâ 1.060 hPa
Nakliye / Saklama koşulları -20 ilâ +60 °C / %10 ilâ 95 BN / 700 ilâ 1.060 hPa
BoyutlarYakl. 140 [G] x 60 [Y] x 105 [D] mm
Türkçe 22Ağırlık

Yakl. 285 g, piller hariç

Takılan kısımManşon Tip BF

Giriş koruması

Aygıt: IP20

Kullanım ömrüAygıt: 5 yıl (günde altı defa kullanıldığında)Manşon: 2 yıl (günde altı defa kullanıldığında)AC adaptörü: 5 yıl (günde altı defa kullanıldığında)Aksesuar AC adaptörü

Adaptör tansiyon monitörü evdeki bir güç
kaynağına bağlar.

TB-233C

Satın almak için lütfen yerel A&D bayinizebaşvurun.
AC adaptörünün periyodik olarak incelenmesi veya
değiştirilmesi gerekir.
AC adaptörü üzerine basılmış semboller
Semboller
Fonksiyon / Anlamı
Yalnızca iç mekanda kullanım içindir
Sınıf II aygıtı
Termal sigorta
SigortaEC yönergesi aygıtı etiketi
EAC sertifika aygıtı etiketi
AC adaptörü fişinin polaritesi

Aksesuarlar ayrıca satılırManşonKatalog Numarası
CUF-F-LA
CUF-I
CUF-F-A
CUF-F-SAManşon Boyutu
Büyük yetişkin
manşonuGeniş aralıklımanşonYetişkinmanşonuKüçük yetişkin
manşonu

Kol Boyutu
31 cm - 45 cm
22 cm - 42 cm
22 cm - 32 cm
16 cm - 24 cm

AC adaptörü
Katalog Numarası
TB-233C

Not:

Fiş
Tip C

Teknik özellikler önceden haber verilmeksizin değiştirilebilir.
IP sınıflandırması, IEC 60529’a uygun kapaklar tarafından sağlanan
koruma derecesidir. Aygıt, parmak gibi 12 mm ve üstü çaptaki katı
yabancı nesnelere karşı korunur. Bu aygıt, suya karşı dayanıklı değildir.
Türkçe 23

NOT

Türkçe 24‫اﻟﻣﺣﺗوﯾﺎت‬‫اﻟﻌﻣﻼء اﻷﻋزاء ‪2 ........................................................................................‬‬‫ﻣﻼﺣظﺎت أوﻟﯾّﺔ ‪2 ........................................................................................‬‬‫اﻟﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ ‪2 .....................................................................................‬‬‫اﻟﺗﻌرﯾف ﺑﺄﺟزاء اﻟﺟﮭﺎز ‪4 ...............................................................................‬‬‫اﻟرﻣوز ‪5 . ................................................................................................‬‬‫ﻧﻣط اﻟﺗﺷﻐﯾل‪7 ............................................................................................‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس ‪8 ..................................................................................‬‬‫ﺗرﻛﯾب ‪ /‬ﺗﻐﯾﯾر اﻟﺑطﺎرﯾﺎت ‪8 .......................................................................‬‬‫ﺗوﺻﯾل ﺧرطوم اﻟﮭواء ‪9 ..........................................................................‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪9 ..............................................................‬‬‫ﺗوﺻﯾل ﻗﺎﺑس ّ‬‫ﺗﻌدﯾل اﻟﺳﺎﻋﺔ اﻟداﺧﻠﯾﺔ ‪10 ..........................................................................‬‬‫اﺧﺗﯾﺎر اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟﻠرﺑﺎط ‪11 ................................................................‬‬‫ﻛﯾﻔﯾﺔ وﺿﻊ رﺑﺎط اﻟذراع ‪11 ...... ................................................................‬‬‫ﻛﯾﻔﯾﺔ إﺟراء ﻋﻣﻠﯾﺎت ﻗﯾﺎس دﻗﯾﻘﺔ ‪13 ..............................................................‬‬‫اﻟﻘﯾﺎس ‪13 ...........................................................................................‬‬‫ﺑﻌد اﻟﻘﯾﺎس ‪13 ......................................................................................‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ‪14 .......................................................................................‬‬‫اﻟﻘﯾﺎس اﻟﻌﺎدي‪14 ...................................................................................‬‬‫اﻟﻘﯾﺎس ﻣﻊ ﺿﻐط ‪15 ....................................................................... SET‬‬‫إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ ‪15 .. ................................‬‬‫ﻣﻼﺣظﺎت ﻟﻘﯾﺎس دﻗﯾﻖ ‪16 .........................................................................‬‬‫ﻗﯾﺎس ‪16 ............................................................................... TriCheckTM‬‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة ‪18 ........................................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣؤﺷر ‪IHB/AFib‬؟ ‪19 . ................................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑـ ‪AFib‬؟ ‪19 ..............................................................................‬‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط ‪19 ................................................................................‬‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ‪19 .............................................................‬‬‫ﻧُﺑذة ﻋن ﻣﺳﺗوى ﺿﻐط اﻟدم ‪20 .........................................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣﺳﺗوى ﺿﻐط اﻟدم؟ ‪20 .............................................................‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎرﺗﻔﺎع ﺿﻐط اﻟدم وﻛﯾف ﯾﻣﻛن اﻟﺗﺣﻛم ﻓﯾﮫ؟ ‪20 .... ................................‬‬‫ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟ ‪20 ................................................‬‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم ‪20 .........................................‬‬‫اﻻﺧﺗﻼﻓﺎت ﻓﻲ ﺿﻐط اﻟدم ‪20 ..... ................................................................‬‬‫اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ ‪21 ......................................................................‬‬‫اﻟﺻﯾﺎﻧﺔ ‪22 ...............................................................................................‬‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻔﻧﯾﺔ ‪22 ........................................................................................‬‬‫اﻟﻌرﺑﯾﺔ ‪1‬‬‫اﻟﻌﻣﻼء اﻷﻋزاء‬‫ﺗﮭﺎﻧﯾﻧﺎ ﻟﻛم ﻋﻠﻰ ﺷراء أﺣدث ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻣن ‪ ،A&D‬اﻟذي ﯾﻌﺗﺑر أﺣد اﻷﺟﮭزة اﻷﻛﺛر‬‫ﺻ ّﻣم ھذا اﻟﺟﮭﺎز ﻟﯾوﻓر ﺳﮭوﻟﺔ ﻓﻲ اﻻﺳﺗﺧدام ودﻗﺔً ﻓﻲ اﻟﻘﯾﺎس‪ ،‬ﺣﯾث ﯾﺳ ّﮭل ﻋﻠﯾك‬‫ﺗطو ًرا ﻓﻲ اﻟوﻗت اﻟﺣﺎﻟﻲ‪ُ .‬‬‫ّ‬‫اﻟﻘﯾﺎس اﻟﯾوﻣﻲ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﻧُوﺻﯾﻛم ﺑﻘراءة ھذا اﻟدﻟﯾل ﺑﻌﻧﺎﯾﺔ ﻗﺑل اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻠﻣرة اﻷوﻟﻰ‪.‬‬‫ﻣﻼﺣظﺎت أوﻟﯾّﺔ‬‫ﯾﺗطﺎﺑﻖ ھذا اﻟﺟﮭﺎز ﻣﻊ اﻟﻣواﺻﻔﺎت اﻷوروﺑﯾﺔ ‪ 93/42 EEC‬اﻟﺧﺎﺻﺔ ﺑﺎﻷﺟﮭزة اﻟطﺑﯾﺔ‪ .‬وﯾﺷﯾر رﻣز‬‫إﻟﻰ دﻗﺔ اﻟﺗطﺎﺑﻖ‪.‬‬‫)‪ :0123‬اﻟرﻗم اﻟﻣرﺟﻌﻲ ﻟﻠﮭﯾﺋﺔ اﻟﻣﻌﻧﯾﺔ اﻟﺗﻲ ﺗم إﺧطﺎرھﺎ(‬‫ﺻ ّﻣم ھذا اﻟﺟﮭﺎز ﻟﻼﺳﺗﺧدام ﻣﻊ اﻟﻛﺑﺎر دون اﻷطﻔﺎل اﻟﺻﻐﺎر وﺣدﯾﺛﻲ اﻟوﻻدة‪.‬‬‫ُ‬‫ﻣﻛﺎن اﻻﺳﺗﺧدام‪ .‬ھذا اﻟﺟﮭﺎز ُﻣﺧﺻص ﻟﻼﺳﺗﻌﻣﺎل ﻷﻏراض اﻟرﻋﺎﯾﺔ اﻟﺻﺣﯾﺔ اﻟﻣﻧزﻟﯾﺔ‪.‬‬‫ﺻﻣم ھذا اﻟﺟﮭﺎز ﻟﻘﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم وﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟﻺﻧﺳﺎن ﻷﻏراض اﻟﺗﺷﺧﯾص‪.‬‬‫ُ‬‫اﻟﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ‬‫ﻛوﻧﺎت دﻗﯾﻘﺔ؛ ﻟذا ﺗﺟﻧّب ﻣطﻠﻘًﺎ ﺗﻌرﯾﺿﮫ ﻟدرﺟﺎت اﻟﺣرارة واﻟرطوﺑﺔ اﻟﻣرﺗﻔﻌﺔ‬‫ُ‬‫ﺻﻧّﻊ ھذا اﻟﺟﮭﺎز ﺑﺎﺳﺗﺧدام ُﻣ ّ‬‫وأﺷﻌﺔ اﻟﺷﻣس اﻟﻣﺑﺎﺷرة واﻟﺻدﻣﺎت واﻟﻐﺑﺎر‪.‬‬‫ّ‬‫ﻧظف اﻟﺟﮭﺎز واﻟرﺑﺎط ﺑﺎﺳﺗﺧدام ﻗطﻌﺔ ﻗﻣﺎش ﺟﺎﻓﺔ وﻧﺎﻋﻣﺔ أو ﻗطﻌﺔ ﻗﻣﺎش ﻣﺑﻠﻠﺔ ﺑﺎﻟﻣﺎء وﻣﻧظف ﻣﻌﺗدل‪.‬‬‫ﺗﺟﻧب ﻣطﻠﻘًﺎ اﺳﺗﺧدام اﻟﻛﺣول أو اﻟﺑﻧزﯾن أو اﻟﺛﻧر أو ﻏﯾرھﺎ ﻣن اﻟﻣواد اﻟﻛﯾﻣﯾﺎﺋﯾﺔ اﻟﻘﺎﺳﯾﺔ ﻣن أﺟل ﺗﻧظﯾف‬‫اﻟﺟﮭﺎز أو اﻟرﺑﺎط‪.‬‬‫ﻲ اﻟرﺑﺎط ﺑﺷدة أو ﺗﺧزﯾن اﻟﺧرطوم ﺑﺛﻧﯾﮫ ﺑﺷدة ﻟﻔﺗرات طوﯾﻠﺔ؛ ﯾؤدي ھذا اﻟﺗﺻرف إﻟﻰ ﻗﺻر ﻣدة‬‫ﺗﺟﻧب ط ّ‬‫اﻟﻣﻛوﻧﺎت‪.‬‬‫اﺳﺗﺧدام ﺗﻠك‬‫ّ‬‫ﺗو ﱠخ اﻟﺣذر ﻟﺗﻔﺎدي أي ﺣوادث اﺧﺗﻧﺎق ﻋن طرﯾﻖ اﻟﺧطﺄ ﻟﻸطﻔﺎل واﻟرﺿﻊ ﻋن طرﯾﻖ ھذا اﻟﺧرطوم أو‬‫اﻟﺳﻠك‪.‬‬‫ﻧظرا‬‫ﻻ ﺗﻘم ﺑﺛﻧﻲ ﺧرطوم اﻟﮭواء أﺛﻧﺎء إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻟﺗﻌرض ﻹﺻﺎﺑﺔ‬‫ً‬‫ﻟﻠﺿﻐط اﻟﻣﺳﺗﻣر ﻋﻠﻰ اﻟرﺑﺎط‪.‬‬‫ﺗﻌرض اﻟﺟﮭﺎز واﻟرﺑﺎط ﻟﻠﺑﻠل ﺑﻔﻌل اﻷﻣطﺎر واﻟﻌرق واﻟﻣﯾﺎه‪.‬‬‫اﻟﺟﮭﺎز واﻟرﺑﺎط ﻏﯾر ﻣﻘﺎوﻣﯾن ﻟﻠﻣﺎء‪ .‬ﺗﺟﻧﱠب ّ‬‫ﯾﻣﻛن أن ﺗﺗﻌرض ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻠﺗﺷوﯾش إذا ﺟرى اﺳﺗﺧدام اﻟﺟﮭﺎز ﺑﺟوار أﺟﮭزة اﻟﺗﻠﻔزﯾون أو‬‫اﻟﻣﯾﻛرووﯾف أو اﻟﮭواﺗف اﻟﻣﺣﻣوﻟﺔ أو اﻷﺟﮭزة اﻟﺗﻲ ﯾﻧﺑﻌث ﻣﻧﮭﺎ اﻷﺷﻌﺔ اﻟﺳﯾﻧﯾﺔ أو اﻷﺧرى ذات اﻟﻣﺟﺎﻻت‬‫اﻟﻛﮭرﺑﺎﺋﯾﺔ اﻟﻘوﯾﺔ‪.‬‬‫أﺟﮭزة اﻻﺗﺻﺎﻻت اﻟﻼﺳﻠﻛﯾﺔ‪ ،‬ﻣﺛل أﺟﮭزة اﻻﺗﺻﺎل اﻟﺷﺑﻛﻲ اﻟﻣﻧزﻟﯾﺔ‪ ،‬واﻟﮭواﺗف اﻟﻣﺣﻣوﻟﺔ‪ ،‬واﻟﮭواﺗف‬‫اﻟﻼﺳﻠﻛﯾﺔ واﻟﻣﺣطﺎت اﻟﻘﺎﻋدﯾﺔ ﻟﮭﺎ‪ ،‬واﻷﺟﮭزة اﻟﻼﺳﻠﻛﯾﺔ اﻟﺗﻲ ﺗﻌﻣل ﺑﻣوﺟﺎت اﻟرادﯾو‪ ،‬ﺟﻣﯾﻌﮭﺎ ﯾﻣﻛن أن ﺗؤﺛر‬‫ﻋﻠﻰ ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫وﻟذﻟك‪ ،‬ﯾﺟب اﻟﺑﻘﺎء ﻋﻠﻰ ﻣﺳﺎﻓﺔ ﻻ ﺗﻘل ﻋن ‪ 30‬ﺳم ﻣن ھذه اﻷﺟﮭزة‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪2‬‬‫اﻷﺟﮭزة واﻷﺟزاء واﻟﺑطﺎرﯾﺎت اﻟﻣﺳﺗﻌﻣﻠﺔ ﻻ ﺗ ُﻌﺎﻣل ﻣﻌﺎﻣﻠﺔ اﻟﻧﻔﺎﯾﺎت اﻟﻣﻧزﻟﯾﺔ اﻟﻌﺎدﯾﺔ‪ ،‬وﯾﺟب اﻟﺗﺧﻠص ﻣﻧﮭﺎ‬‫وﻓﻘًﺎ ﻟﻠواﺋﺢ اﻟﻣﺣﻠﯾﺔ اﻟﻣﻌﻣول ﺑﮭﺎ‪.‬‬‫اﻟﻣﺣول ﻣن اﻟﻣﺧرج اﻟﻛﮭرﺑﺎﺋﻲ ﻋﻧد‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪ ،‬ﺗﺄ ﱠﻛد ﻣن ﺳﮭوﻟﺔ إﻣﻛﺎﻧﯾﺔ إزاﻟﺔ‬‫ﻋﻧد اﺳﺗﺧدام‬‫ّ‬‫ّ‬‫اﻟﺿرورة‪.‬‬‫ﻋﻧد إﻋﺎدة اﺳﺗﺧدام اﻟﺟﮭﺎز‪ ،‬ﺗﺄﻛد ﻣن أﻧﮫ ﻧظﯾف‪.‬‬‫ﻻ ﺗﻌدّل ﻓﻲ اﻟﺟﮭﺎز‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ وﻗوع ﺣوادث أو ﺗﻠف اﻟﺟﮭﺎز‪.‬‬‫ﯾﺗطﻠب ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬اﻟﺿﻐط ﻋﻠﻰ اﻟذراع ﻣن ﺧﻼل اﻟرﺑﺎط ﺑﻘوة ﺗﻛﻔﻲ ﻹﯾﻘﺎف ﺗدﻓﻖ اﻟدم ﻣؤﻗﺗ ًﺎ‬‫ﻓﻲ ﺷرﯾﺎن اﻟذراع‪ .‬ﯾﻣﻛن أن ﯾُﺳﺑّب ذﻟك ﻓﻲ اﻟﺷﻌور ﺑﺄﻟم أو ﺗﻧﻣﯾل أو ظﮭور ﻋﻼﻣﺔ ﺣﻣراء ﻣؤﻗﺗﺔ ﺑﺎﻟذراع‪.‬‬‫ﺗظﮭر ھذه اﻟﻌﻼﻣﺎت ﻻ ﺳﯾﻣﺎ ﻋﻧد ﺗﻛرار ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس ﺑﺷﻛل ﻣﺗﺗﺎﺑﻊ‪.‬‬‫ﺳﯾﺧﺗﻔﻲ أي أﻟم أو ﺗﻧﻣﯾل أو ﻋﻼﻣﺎت ﺣﻣراء ﺑﻣرور اﻟوﻗت‪.‬‬‫ﯾﻣﻛن أن ﯾﺗﺳﺑب ﺗﻛرار ﻋﻣﻠﯾﺎت ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﺑﺷﻛل ﻣﺗﺗﺎﺑﻊ‪ ،‬ﻓﻲ ﺣدوث ﻣﺷﻛﻠﺔ ﺻﺣﯾﺔ ﺑﺳﺑب‬‫ﺗﻛرار اﻟﺿﻐط ﻋﻠﻰ ﺗدﻓﻖ اﻟدم‪ .‬ﺗﺄﻛد ﻣن أن ﻋﻣل اﻟﺟﮭﺎز ﻻ ﯾؤدي إﻟﻰ ﺿﻌف طوﯾل اﻷﻣد ﻓﻲ اﻟدورة‬‫اﻟدﻣوﯾﺔ‪ ،‬ﻋﻧد اﺳﺗﺧدام اﻟﺟﮭﺎز ﺑﺎﺳﺗﻣرار‪.‬‬‫ت ﻋﻣﻠﯾﺔ اﺳﺗﺋﺻﺎل اﻟﺛدي‪ ،‬ﻓﺎﺳﺗﺷﯾري اﻟطﺑﯾب ﻗﺑل اﺳﺗﺧدام اﻟﺟﮭﺎز‪.‬‬‫إذا ﻛﻧت ﻗد أﺟرﯾ ِ‬‫ﻻ ﺗﺳﻣﺢ ﻟﻸطﻔﺎل ﺑﺎﺳﺗﺧدام اﻟﺟﮭﺎز ﻣن ﺗﻠﻘﺎء أﻧﻔﺳﮭم واﺣرص ﻋﻠﻰ أن ﯾﻛون اﺳﺗﺧدام اﻟﺟﮭﺎز ﻓﻲ ﻣﻛﺎن ﺑﻌﯾد‬‫ﻋن ﻣﺗﻧﺎول اﻷطﻔﺎل‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ وﻗوع ﺣوادث أو ﺗﻠف‪.‬‬‫ﺛﻣﺔ أﺟزاء ﺻﻐﯾرة ﻓﻲ اﻟﺟﮭﺎز ﯾﻣﻛن أن ﺗﺳﺑب اﻻﺧﺗﻧﺎق ﻟﻸطﻔﺎل ﻓﻲ ﺣﺎل اﺑﺗﻼﻋﮭﺎ ﻋن طرﯾﻖ اﻟﺧطﺄ‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻋﻧد ﻋدم اﺳﺗﺧداﻣﮫ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﻓﺻل ّ‬‫ﯾﻌرض اﺳﺗﺧدام ﻣﻠﺣﻘﺎت ﻻ ﺗرد ﺗﻔﺻﯾﻼً ﻓﻲ ھذا اﻟدﻟﯾل ﺳﻼﻣﺗك ﻟﻠﺧطر‪.‬‬‫ﻗد ّ‬‫إذا ﺗﺳﺑﺑت اﻟﺑطﺎرﯾﺔ ﻓﻲ ﺣدوث داﺋرة ﻗﺻر‪ ،‬ﻓﯾﻣﻛن أن ﺗﺻﺑﺢ ﺳﺎﺧﻧﺔ وﺗﺗﺳﺑب ﻓﻲ ﺣروق ﺑﺎﻟﺟﻠد‪.‬‬‫اﺗرك اﻟﺟﮭﺎز ﯾﺗﻛﯾّف ﻋﻠﻰ اﻟﺑﯾﺋﺔ اﻟﻣﺣﯾطﺔ ﻗﺑل اﻻﺳﺗﺧدام )ﻣدة ﺳﺎﻋﺔ واﺣدة ﺗﻘرﯾﺑًﺎ(‪.‬‬‫ﻟم ﺗُﺟر اﺧﺗﺑﺎرات ﺳرﯾرﯾﺔ ﻋﻠﻰ اﻷطﻔﺎل ﺣدﯾﺛﻲ اﻟوﻻدة واﻟﻧﺳﺎء اﻟﺣواﻣل‪ .‬ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻋﻠﻰ اﻷطﻔﺎل‬‫ﺣدﯾﺛﻲ اﻟوﻻدة أو اﻟﻧﺳﺎء اﻟﺣواﻣل‪.‬‬‫اﻟﺗﻌرض‬‫ﻻ ﺗﻠﻣس اﻟﺑطﺎرﯾﺎت وﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر واﻟﻣرﯾض ﺳوﯾًﺎ ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ‪ .‬ﻗد ﯾؤدي ذﻟك إﻟﻰ‬‫ّ‬‫ﻟﺻدﻣﺔ ﻛﮭرﺑﺎﺋﯾﺔ‪.‬‬‫ﻻ ﺗﺑدأ ﻓﻲ ﻧ ْﻔﺦ اﻟﮭواء دون ﻟف اﻟرﺑﺎط ﺣول أﻋﻠﻰ اﻟذراع )اﻟﻌﺿد(‪.‬‬‫ﻣواﻧﻊ اﻻﺳﺗﻌﻣﺎل‬‫ﻓﯾﻣﺎ ﯾﻠﻲ اﺳﺗﻌراض ﻟﻠﺗداﺑﯾر اﻻﺣﺗﯾﺎطﯾﺔ ﻣن أﺟل اﺳﺗﺧدام ﻣﻼﺋم ﻟﻠﺟﮭﺎز‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﯾﺗﺻل ﺑﮫ ﺟﮭﺎز طﺑﻲ ﻛﮭرﺑﺎﺋﻲ آﺧر‪ .‬ﻗد ﻻ ﯾﻌﻣل اﻟﺟﮭﺎز ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﯾﺟب ﻋﻠﻰ اﻷﺷﺧﺎص اﻟذﯾن ﯾﻌﺎﻧون ﻣن ﻧﻘص ﺷدﯾد ﻓﻲ اﻟدورة اﻟدﻣوﯾﺔ ﻓﻲ اﻟذراع اﺳﺗﺷﺎرة اﻟطﺑﯾب ﻗﺑل‬‫اﻟﺗﻌرض ﻟﻠﻣﺷﻛﻼت اﻟطﺑﯾﺔ‪.‬‬‫اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﺗﻔﺎدي‬‫ّ‬‫ﻻ ﺗﺷ ّﺧص ﺑﻧﻔﺳك ﻧﺗﺎﺋﺞ اﻟﻘﯾﺎس وﺗﺑدأ ﻓﻲ ﺗﻧﺎول اﻟﻌﻼج ﻣن ﺗﻠﻘﺎء ﻧﻔﺳك‪ .‬اﺳﺗﺷر اﻟطﺑﯾب داﺋ ًﻣﺎ ﻣن أﺟل ﺗﻘﯾﯾم‬‫اﻟﻧﺗﺎﺋﺞ واﻟﻌﻼج‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﺑﮫ ﺟرح ﻟم ﯾﻠﺗﺋم ﺑﻌد‪.‬‬‫ﻻ ﺗﻠف اﻟرﺑﺎط ﺣول اﻟذراع اﻟذي ﯾﺗﻠﻘﻰ اﻟﺗﻧﻘﯾط اﻟورﯾدي أو ﻧﻘل اﻟدم‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻟﺗﻌرض‬‫ﻟﻺﺻﺎﺑﺔ أو اﻟﺿرر‪.‬‬‫ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻓﻲ أﻣﺎﻛن وﺟود اﻟﻐﺎزات اﻟﻘﺎﺑﻠﺔ ﻟﻼﺷﺗﻌﺎل‪ ،‬ﻣﺛل ﻏﺎزات اﻟﺗﺧدﯾر‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك‬‫ﻓﻲ ﺣدوث اﻧﻔﺟﺎر‪.‬‬‫ﻻ ﺗﺳﺗﺧدم اﻟﺟﮭﺎز ﻓﻲ أﻣﺎﻛن ﺗرﻛﯾز اﻷﻛﺳﺟﯾن ﻓﯾﮭﺎ ﻣرﺗﻔﻊ‪ ،‬ﻣﺛل ﻏرﻓﺔ اﻷﻛﺳﺟﯾن ﻋﺎﻟﯾﺔ اﻟﺿﻐط أو ﺧﯾﻣﺔ‬‫اﻷﻛﺳﺟﯾن‪ .‬ﯾﻣﻛن أن ﯾﺗﺳﺑب ذﻟك ﻓﻲ اﻧدﻻع ﺣرﯾﻖ أو ﺣدوث اﻧﻔﺟﺎر‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪3‬‬‫اﻟﺗﻌرﯾف ﺑﺄﺟزاء اﻟﺟﮭﺎز‬‫ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫اﻟﺷﺎﺷﺔ‬‫زر‬‫زر‬‫زر ‪SET‬‬‫ﺻل اﻟﮭواء‬‫ﻗﺎﺑس ﻣو ّ‬‫اﻟزر ‪START‬‬‫ﻣﻘﺑس اﻟﮭواء‬‫ﺧرطوم اﻟﮭواء‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫رﻣز ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫رﺑﺎط اﻟذراع‬‫رﻣز اﻟﻣؤﺷر‬‫ﻏطﺎء اﻟﺑطﺎرﯾﺎت‬‫ﺣﺟرة اﻟﺑطﺎرﯾﺎت‬‫ﺑطﺎرﯾﺎت ‪ 1.5‬ﻓوﻟت‬‫)‪ R6P‬أو ‪ LR6‬أو ‪(AA‬‬‫إﻋدادات اﻟﺿﻐط‬‫ﻣﺗوﺳط‬‫اﻟذاﻛرة‬‫اﻟﺷﺎﺷﺔ‬‫رﻣز ‪IHB/AFib‬‬‫رﻣز ﺧطﺄ ﻓﻲ ﺗﺛﺑﯾت اﻟرﺑﺎط‬‫اﻟﺿﻐط‬‫اﻷﻧﻘﺑﺎﺿﻲ‬‫رﻣز ﺧطﺄ اﻟﺣرﻛﺔ‬‫رﻣز ‪TriCheckTM‬‬‫اﻟﺿﻐط‬‫اﻻﻧﺑﺳﺎطﻲ‬‫رﻣز اﻟﻘﻠب‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‬‫ﻣؤﺷر اﻟﺑطﺎرﯾﺔ‬‫ﺳرﻋﺔ اﻟﻧﺑض‬‫ﻋرض اﻟﺳﺎﻋﺔ‬‫رﻣز ‪PM / AM‬‬‫اﻟﻌرﺑﯾﺔ ‪4‬‬

‫اﻟرﻣوز‬‫وﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ ﻋﻠﺑﺔ اﻟﺟﮭﺎز‬‫ّ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫اﻟرﻣوز‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫وﺿﻊ اﻻﺳﺗﻌداد وﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫دﻟﯾل ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت‬‫ﺗﯾﺎر ﻣﺑﺎﺷر‬‫‪SN‬‬‫اﻟرﻗم اﻟﺗﺳﻠﺳﻠﻲ‬‫ﺗﺎرﯾﺦ اﻟﺗﺻﻧﯾﻊ‬‫اﻟﻧوع ‪ :BF‬ﺗم ﺗﺻﻣﯾم اﻟﺟﮭﺎز‪ ،‬واﻟرﺑﺎط‪ ،‬واﻷﻧﺑوب‬‫ﻟﺗوﻓﯾر ﺣﻣﺎﯾﺔ ﺧﺎﺻﺔ ﻣن اﻟﺻدﻣﺎت اﻟﻛﮭرﺑﺎﺋﯾﺔ‪.‬‬‫ﻣﻠﺻﻖ اﻟﺗوﺟﯾﮫ ‪ EC‬ﻟﻸﺟﮭزة اﻟطﺑﯾﺔ‬‫رﻣز اﻟدﻟﯾل اﻟﺗوﺟﯾﮭﻲ ﻟﻧﻔﺎﯾﺎت اﻟﻣﻌدات اﻟﻛﮭرﺑﺎﺋﯾﺔ‬‫واﻹﻟﻛﺗروﻧﯾﺔ‬‫اﻟﺟﮭﺔ اﻟﻣﺻﻧّﻌﺔ‬‫ﻣﻣﺛل اﻻﺗﺣﺎد اﻷوروﺑﻲ‬‫ﻗطﺑﯾﺔ ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫‪IP‬‬‫رﻣز اﻟﺣﻣﺎﯾﺔ اﻟدوﻟﯾﺔ‬‫ﺣﺎﻓظ ﻋﻠﯾﮫ ﺟﺎﻓًﺎ‬‫اﻟرﻣوز اﻟﺗﻲ ﺗظﮭر ﻋﻠﻰ اﻟﺷﺎﺷﺔ‬‫اﻟرﻣوز‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﯾظﮭر ﺣﯾﻧﻣﺎ ﯾﻛون إﺟراء اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬ﯾوﻣض‬‫ﻋﻧد اﻟﻛﺷف ﻋن اﻟﻧﺑض‪.‬‬‫رﻣز ‪IHB/AFib‬‬‫ﯾظﮭر ﻋﻧد اﻟﻛﺷف ﻋن ﻋدم اﻧﺗظﺎم ﻓﻲ ﺿرﺑﺎت‬‫اﻟﻘﻠب‪.‬‬‫ﯾﻣﻛن أن ﯾُﺿﻲء ﻋﻧد اﻟﻛﺷف ﻋن وﺟود اھﺗزاز ﺑﺳﯾط‬‫ﻟﻠﻐﺎﯾﺔ‪ ،‬ﻣﺛل اﻻھﺗزاز أو اﻻرﺗﻌﺎش‪.‬‬‫ﻧﻣط ‪TriCheckTM‬‬‫ﯾﺄﺧذ ﺛﻼﺛﺔ ﻗﯾﺎﺳﺎت ﻣﺗﺗﺎﻟﯾﺔ وﯾﻌرض أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‬‫ﻣﺗوﺳط ﻗﯾّم اﻟﻘﯾﺎﺳﺎت اﻟﺛﻼﺛﺔ‪.‬‬‫ﺗﺣرك اﻟﺟﺳم أو اﻟذراع أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ﯾظﮭر ﻋﻧد ّ‬‫ﯾظﮭر ﻓﻲ أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس ﻓﻲ ﺣﺎل ﻋدم ﻟف‬‫اﻟرﺑﺎط ﺑﺈﺣﻛﺎم ﺣول اﻟذراع‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪5‬‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫إﺟراء اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‪ .‬ﯾظل ﺛﺎﺑﺗًﺎ‬‫ﻗدر اﻟﻣﺳﺗطﺎع‪.‬‬‫ﯾﻣﻛن أن ﯾؤدي إﻟﻰ إظﮭﺎر ﻗﯾﻣﺔ‬‫ﺧطﺄ‪ .‬ﻗم ﺑﺈﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ُﻛن ﺛﺎﺑﺗًﺎ أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﯾﻣﻛن أن ﯾؤدي إﻟﻰ إظﮭﺎر ﻗﯾﻣﺔ‬‫ﺧطﺄ‪ .‬ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪،‬‬‫وﻗم ﺑﺈﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫اﻟرﻣوز‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس اﻟﺳﺎﺑﻘﺔ اﻟ ُﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‪.‬‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫ﻣﺗوﺳط اﻟﺑﯾﺎﻧﺎت )ﻣﺗوﺳط ﻋدد اﻟﻘﯾﺎﺳﺎت اﻟﻣﺧزﻧﺔ(‬‫اﻟﺑطﺎرﯾﺔ ﻣﻣﺗﻠﺋﺔ‬‫ﻣؤﺷر طﺎﻗﺔ اﻟﺑطﺎرﯾﺔ ﻓﻲ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى‬‫ﺗﻛون طﺎﻗﺔ اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔﺿﺔ ﻋﻧدﻣﺎ ﯾوﻣض اﻟرﻣز‪ .‬ﺟدﯾدة ﻋﻧدﻣﺎ ﯾوﻣض اﻟرﻣز‪.‬‬‫ﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة‬‫اﻟﺗﺣرك ﻓﻲ أﺛﻧﺎء‬‫ﻧظرا إﻟﻰ‬‫ّ‬‫ﺿﻐط اﻟدم ﻏﯾر ﻣﺳﺗﻘر ً‬‫أﺧرى‪.‬‬‫إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ُﻛن ﺛﺎﺑﺗًﺎ أﺛﻧﺎء إﺟراء اﻟﻘﯾﺎس‪.‬‬‫ﻗﯾم اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ واﻻﻧﺑﺳﺎطﻲ ﺿﻣن ﻧطﺎق‬‫‪ mmHg 10‬ﻣن ﻛﻠﯾﮭﻣﺎ‪.‬‬‫ﻗﯾﻣﺔ اﻟﺿﻐط ﻻ ﺗرﺗﻔﻊ أﺛﻧﺎء ﻧ ْﻔﺦ اﻟﮭواء‪.‬‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪،‬‬‫وﺣﺎول إﻋﺎدة إﺟراء اﻟﻘﯾﺎس ﻣرة‬‫ﻟم ﯾُﻠف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫أﺧرى‪.‬‬‫‪.PUL DISPLAY ERROR‬‬‫ﻟم ﯾﺗم اﻛﺗﺷﺎف ﺿرﺑﺎت اﻟﻘﻠب ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫ﺧطﺄ داﺧﻠﻲ ﻓﻲ ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫‪SYS‬‬‫‪DIA‬‬‫‪PUL‬‬‫‪AM‬‬‫‪PM‬‬‫ﺿﻐط اﻟدم اﻻﻧﻘﺑﺎﺿﻲ ﺑـ ‪mmHg‬‬‫ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ ﺑـ ‪mmHg‬‬‫ﻋدد اﻟﺿرﺑﺎت ﻓﻲ اﻟدﻗﯾﻘﺔ‬‫اﻟﺑﯾﺎﻧﺎت ُﻣﺳ ّﺟﻠﺔ ﻓﻲ اﻟﻔﺗرة ﻣن اﻟﺳﺎﻋﺔ ‪ 4:00‬إﻟﻰ‬‫اﻟﺳﺎﻋﺔ ‪9:59‬‬‫اﻟﺑﯾﺎﻧﺎت ُﻣﺳ ّﺟﻠﺔ ﻓﻲ اﻟﻔﺗرة ﻣن اﻟﺳﺎﻋﺔ ‪ 18:00‬إﻟﻰ‬‫اﻟﺳﺎﻋﺔ ‪1:59‬‬‫إﻋدادات اﻟﺿﻐط‬‫ﯾُﺑﯾن ﻗﯾﻣﺔ اﻟﺿﻐط اﻟﺗﻲ ﺣددھﺎ اﻟﻣﺳﺗﺧدم ﺳﺎﺑﻘًﺎ‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪6‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت واﺿﻐط ﻋﻠﻰ اﻟزر‬‫‪ ، START‬ﺛم أ ِﻋد ﺗرﻛﯾب‬‫اﻟﺑطﺎرﯾﺎت ﻣرة أﺧرى‪ .‬اﺗﺻل‬‫ﺑﺎﻟوﻛﯾل إذا اﺳﺗﻣر ظﮭور اﻟﺧطﺄ‪.‬‬‫ﻧﻣط اﻟﺗﺷﻐﯾل‬‫‪ .1‬اﻟﻘﯾﺎس اﻟﻌﺎدي‬‫اﺿﻐط ﻋﻠﻰ زر ‪ . START‬ﺗﺑدأ ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم وﺗُﺳ ّﺟل اﻟﺑﯾﺎﻧﺎت ﻓﻲ داﺧل اﻟﺟﮭﺎز‪ .‬ﯾﻣﻛن‬‫ﻟﻠﺟﮭﺎز ﺗﺳﺟﯾل آﺧر ﺗﺳﻌﯾن ﻣﺟﻣوﻋﺔ ﻣن اﻟﺑﯾﺎﻧﺎت ﻓﻲ اﻟذاﻛرة‬‫‪ .2‬اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت‬‫أو زر ﻻﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت اﻟﻣوﺟودة ﻓﻲ‬‫اﺿﻐط ﻋﻠﻰ‬‫اﻟذاﻛرة‪ .‬ﯾُﻌرض اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻛل ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‪ ،‬ﻛﻣﺎ ھو‬‫ُﻣﺣدد ﻓﻲ اﻟﺷﻛل اﻟﻣوﺟود ﻋﻠﻰ اﻟﯾﻣﯾن‪.‬‬‫ﺑﻌد ذﻟك‪ ،‬ﻓﻲ ﻛل ﻣرة ﯾﺗم ﻓﯾﮭﺎ‬‫ﺑﯾﺎﻧﺎت اﻟذاﻛرة ﻛﺎﻟﺗﺎﻟﻲ‪.‬‬‫اﻟﺿﻐط ﻋﻠﻰ اﻟزر‪ ،‬ﺗﺳﺗﻌرض‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻛل اﻟﻘﯾﺎﺳﺎت اﻟﺻﺑﺎﺣﯾﺔ اﻟﻣﺄﺧوذة ﺑﯾن‬‫اﻟﺳﺎﻋﺔ ‪ 4:00‬و‪.9:59‬‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻛل اﻟﻘﯾﺎﺳﺎت اﻟﻣﺳﺎﺋﯾﺔ اﻟﻣﺄﺧوذة ﺑﯾن اﻟﺳﺎﻋﺔ‬‫‪ 18:00‬و‪.1:59‬‬‫أﺣدث اﻟﺑﯾﺎﻧﺎت )اﻟﻌدد‪ ،‬ﻋﻠﻰ ﺳﺑﯾل اﻟﻣﺛﺎل‪ ،‬رﻗم ‪(35‬‬‫آﺧر اﻟﺑﯾﺎﻧﺎت )رﻗم ‪(1‬‬‫ﻟﺗﻔﺎﺻﯾل ﻋن اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت‪ ،‬راﺟﻊ ﻗﺳم " اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن‬‫اﻟذاﻛرة"‪) .‬راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪(.18‬‬‫‪ .3‬ﻣﺳﺢ ﻛل اﻟﺑﯾﺎﻧﺎت اﻟ ُﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‬‫زري‬‫اﺿﻐط ﻋﻠﻰ ﻛل ﻣن َ‬‫و‬‫‪ .‬ﯾظﮭر رﻣز‬‫وﻣؤﺷر‬‫زري و ﺣﺗﻰ ﯾﺑدأ اﻟرﻣز‬‫اﻟﺑطﺎرﯾﺔ‪ .‬اﺿﻐط ﻋﻠﻰ ﻛل ﻣن َ‬‫اﻟﻣﺿﯾﺊ ﻓﻲ اﻟوﻣﯾض ﻟﻣﺳﺢ ﻛل اﻟﺑﯾﺎﻧﺎت اﻟ ُﻣﺧزﻧﺔ ﻓﻲ اﻟذاﻛرة‪.‬‬‫‪ .4‬إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ‬‫راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪ 15‬ﻟﻠﺗﻌرف ﻋﻠﻰ إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣطﻠوب‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪7‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﺗرﻛﯾب ‪ /‬ﺗﻐﯾﯾر اﻟﺑطﺎرﯾﺎت‬‫‪ .1‬اﻧزع ﻏطﺎء اﻟﺑطﺎرﯾﺎت‪.‬‬‫‪ .2‬اﻧزع اﻟﺑطﺎرﯾﺎت اﻟﻣﺳﺗﺧدﻣﺔ وأدﺧل اﻷﺧرى‬‫اﻟﺟدﯾدة ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت ﻛﻣﺎ ھو ﻣوﺿّﺢ‪ ،‬ﻣﻊ‬‫ﻣراﻋﺎة اﻻﺗﺟﺎه اﻟﺻﺣﯾﺢ ﻟﻌﻼﻣﺗﻲ اﻟﻘطﺑﯾﺔ )‪ +‬و ‪.(-‬‬‫ﻻ ﺗﺳﺗﺧدم ﺳوى ﺑطﺎرﯾﺎت ﻣﻘﺎس ‪ R6P‬أو ‪LR6‬‬‫أو ‪.AA‬‬‫‪ .3‬ر ّﻛب ﻏطﺎء اﻟﺑطﺎرﯾﺎت‪.‬‬‫اﻟﺧطوة ‪1‬‬‫اﻟﺧطوة ‪3‬‬‫اﻟﺧطوة ‪1‬‬‫اﻟﺧطوة ‪2‬‬‫ﺗﻧﺑﯾﮫ‬‫أدﺧل اﻟﺑطﺎرﯾﺎت اﻟﺟدﯾدة ﻛﻣﺎ ھو ﻣوﺿّﺢ ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت‪ .‬ﻟن ﯾﻌﻣل اﻟﺟﮭﺎز ﻓﻲ ﺣﺎﻟﺔ ﺗرﻛﯾﺑﮭﺎ ﺑطرﯾﻘﺔ‬‫ﻏﯾر ﺻﺣﯾﺣﺔ‪.‬‬‫)رﻣز ‪ (LOW BATTERY‬ﻋﻠﻰ اﻟﺷﺎﺷﺔ‪ ،‬اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪ .‬ﻻ‬‫ﻋﻧدﻣﺎ ﯾوﻣض‬‫ﺗﻌرض‬‫ﺗﺳﺗﺧدم ﺑطﺎرﯾﺎت ﺟدﯾدة ﻣﻊ أﺧرى ﻗدﯾﻣﺔ‪ .‬ﻗد ﯾﺗﺳﺑب ذﻟك ﻓﻲ ﺗﻘﺻﯾر ﻓﺗرة اﺳﺗﺧدام اﻟﺑطﺎرﯾﺎت‪ ،‬أو ﻓﻲ ّ‬‫اﻟﺟﮭﺎز ﻟﻠﻌطل‪.‬‬‫اﺳﺗﺑدل اﻟﺑطﺎرﯾﺎت ﺑﻌد إﯾﻘﺎف ﺗﺷﻐﯾل اﻟﺟﮭﺎز ﺑﺛﺎﻧﯾﺗﯾن أو أﻛﺛر‪.‬‬‫ﻓﺄﺟر ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫)رﻣز ‪ (LOW BATTERY‬ﺣﺗﻰ ﺑﻌد اﺳﺗﺑدال اﻟﺑطﺎرﯾﺎت‪،‬‬‫إذا ظﮭر‬‫ِ‬‫ﯾﺗﻌرف اﻟﺟﮭﺎز ﺑﻌدﺋ ٍذ ﻋﻠﻰ اﻟﺑطﺎرﯾﺎت اﻟﺟدﯾدة‪.‬‬‫ﯾﻣﻛن أن‬‫ّ‬‫ﻻ ﯾظﮭر )رﻣز ‪ (LOW BATTERY‬ﻋﻧدﻣﺎ ﺗﻛون اﻟﺑطﺎرﯾﺎت ﻓﺎرﻏﺔ‪.‬‬‫ﺗﺗﺑﺎﯾن ﻓﺗرة ﺗﺷﻐﯾل اﻟﺑطﺎرﯾﺎت ﺣﺳب درﺟﺔ اﻟﺣرارة اﻟﻣﺣﯾطﺔ‪ ،‬وﻗد ﺗﻛون ﻓﺗرة ﺗﺷﻐﯾﻠﮭﺎ أﻗﺻر ﻋﻧد درﺟﺎت‬‫اﻟﺣرارة اﻟﻣﻧﺧﻔﺿﺔ‪ .‬وﺑﺷﻛل ﻋﺎم‪ ،‬ﺳﺗدوم اﻟﺑطﺎرﯾﺎت اﻷرﺑﻊ اﻟﺟدﯾدة ﻣﻘﺎس ‪ LR6‬ﻣدة ﻋﺎم واﺣد ﺗﻘرﯾﺑًﺎ ﻋﻧد‬‫اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻠﻘﯾﺎس ﻣرﺗﯾن ﻛل ﯾوم‪.‬‬‫ﻻ ﺗﺳﺗﺧدم ﺳوى اﻟﺑطﺎرﯾﺎت اﻟﻣﺣددة ﻓﻘط‪ .‬وﺗﻛون اﻟﺑطﺎرﯾﺎت اﻟﻣرﻓﻘﺔ ﻣﻊ اﻟﺟﮭﺎز ﻷﻏراض اﺧﺗﺑﺎر أداء ﺟﮭﺎز‬‫اﻟﻘﯾﺎس‪ ،‬وﯾﻣﻛن أن ﺗﻛون ذات ﻓﺗرة ﺗﺷﻐﯾل ﻣﺣدودة‪.‬‬‫اﻧزع اﻟﺑطﺎرﯾﺎت ﻓﻲ ﺣﺎﻟﺔ ﻋدم اﺳﺗﺧدام اﻟﺟﮭﺎز ﻟﻔﺗرة طوﯾﻠﺔ‪.‬‬‫ﺗﺗﻌرض اﻟﺑطﺎرﯾﺎت ﻟﻠﺗﺳرﯾب ﻣﻣﺎ ﯾﺗﺳﺑب ﻓﻲ ّ‬‫ﺗﻌطل اﻟﺟﮭﺎز‪.‬‬‫ﯾﻣﻛن أن‬‫ّ‬‫اﻟﻌرﺑﯾﺔ ‪8‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﺗوﺻﯾل ﺧرطوم اﻟﮭواء‬‫ﻣﻘﺑس اﻟﮭواء‬‫ﺻل اﻟﮭواء ﻓﻲ ﻣﻘﺑس اﻟﮭواء‬‫أدﺧل ﻗﺎﺑس ﻣو ّ‬‫ﺑﺈﺣﻛﺎم‪.‬‬‫ﺻل اﻟﮭواء‬‫ﻗﺎﺑس ﻣو ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﺗوﺻﯾل ﻗﺎﺑس‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻓﻲ ﻣﻘﺑس اﻟﺗﯾﺎر‬‫أدﺧل ﻗﺎﺑس‬‫ّ‬‫اﻟﻣﺳﺗﻣر‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﺑﻣﺧرج اﻟﺗﯾﺎر‬‫ﺻل‬‫ﺑﻌد ذﻟك‪ِ ،‬‬‫ّ‬‫اﻟﻛﮭرﺑﺎﺋﻲ‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻗﺎﺑس ّ‬‫ﻣﻘﺑس اﻟﺗﯾﺎر اﻟﻣﺳﺗﻣر‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد اﻟﻣﺣدد‪.‬‬‫اﺳﺗﺧدم ّ‬‫)راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪.(23‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد واﺳﺣﺑﮫ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻣن ﻣﺧرج اﻟﺗﯾﺎر اﻟﻛﮭرﺑﺎﺋﻲ‪ ،‬أﻣﺳك ھﯾﻛل ّ‬‫ﻋﻧد ﻓﺻل ّ‬‫ﺧﺎرج اﻟﻣﺧرج‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ﻋﻧد ﻓﺻل‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد ﻣن ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬أﻣﺳك ﺑﻘﺎﺑس ّ‬‫ّ‬‫واﺳﺣﺑﮫ ﺧﺎرج اﻟﺟﮭﺎز‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪9‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﺗﻌدﯾل اﻟﺳﺎﻋﺔ اﻟداﺧﻠﯾﺔ‬‫اﺿﻐط‬‫ﺗﻌدﯾل اﻟﺳﺎﻋﺔ ﻗﺑل اﻻﺳﺗﺧدام‪.‬‬‫‪ .1‬اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﺣﺗﻰ ﯾﺑدأ اﻟﻌﺎم ﻓﻲ اﻟوﻣﯾض‪.‬‬‫أو‬‫‪ .2‬اﺧﺗر اﻟﻌﺎم ﺑﺎﺳﺗﺧدام زر أو‬‫اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﺗﺣدﯾد اﻟﻌﺎم اﻟﺣﺎﻟﻲ واﻻﻧﺗﻘﺎل إﻟﻰ اﺧﺗﯾﺎر‬‫اﻟﯾوم‪/‬اﻟﺷﮭر‪ .‬ﯾﻣﻛن ﺿﺑط اﻟﺗﺎرﯾﺦ ﻓﻲ أي وﻗت ﺑﯾن ﻋﺎﻣﻲ ‪2010‬‬‫و‪.2059‬‬‫اﻟﺳﻧﺔ‬‫‪.‬‬‫‪ .3‬اﺧﺗر اﻟﺷﮭر ﺑﺎﺳﺗﺧدام زر أو‬‫اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﺗﺣدﯾد اﻟﺷﮭر اﻟﺣﺎﻟﻲ واﻻﻧﺗﻘﺎل إﻟﻰ‬‫اﺧﺗﯾﺎر اﻟﯾوم‪.‬‬‫أو‬‫اﻟﺷﮭر‬

‫‪.‬‬

‫أو‬

‫‪ .4‬اﺧﺗر اﻟﯾوم ﺑﺎﺳﺗﺧدام زر أو‬‫اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﺗﺣدﯾد اﻟﯾوم اﻟﺣﺎﻟﻲ واﻻﻧﺗﻘﺎل إﻟﻰ اﺧﺗﯾﺎر‬‫اﻟﺳﺎﻋﺔ‪/‬اﻟدﻗﯾﻘﺔ‪.‬‬
‫‪.‬‬

‫أو‬
‫اﻟﺳﺎﻋﺔ‬‫‪ .5‬اﺧﺗر اﻟﺳﺎﻋﺔ ﺑﺎﺳﺗﺧدام زر أو‬‫اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﺗﺣدﯾد اﻟﯾوم اﻟﺣﺎﻟﻲ واﻻﻧﺗﻘﺎل إﻟﻰ اﺧﺗﯾﺎر‬‫اﻟدﻗﯾﻘﺔ‪.‬‬‫‪.‬‬‫‪ .6‬اﺧﺗر اﻟدﻗﯾﻘﺔ ﺑﺎﺳﺗﺧدام زر أو‬‫)اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﻠذھﺎب إﻟﻰ إﻋدادات ﻧﻣط اﻟﺿﻐط‪.‬‬‫ﻟﻠﺗﻔﺎﺻﯾل‪ ،‬راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪.(15‬‬‫اﺿﻐط ﻋﻠﻰ زر ‪ START‬ﻹطﻔﺎء اﻟﺟﮭﺎز‪.‬‬‫‪.‬‬‫ﻣﻠﺣوظﺔ‪ :‬ﺑﻌد ﺛﻼث دﻗﺎﺋﻖ ﻣن ﻋدم اﻻﺳﺗﺧدام‪ ،‬ﺳﯾﺗم إطﻔﺎء اﻟﺟﮭﺎز‬‫أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫ﻋﻧد ﻋدم ﺿﺑط اﻟﺳﺎﻋﺔ‪ ،‬ﺗﻌرض ﺷﺎﺷﺔ‬‫اﻟﺳﺎﻋﺔ ﺷرطﺎت ﻛﻣﺎ ھو ُﻣوﺿﺢ ﻋﻠﻰ‬‫اﻟﯾﻣﯾن‪.‬‬‫اﻟﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ زر أو ﺳﯾﻐﯾر اﻟﻘﯾﻣﺔ‬‫ﺑﺎﺳﺗﻣرار‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪10‬‬‫اﻟﯾوم‬‫أو‬‫اﻟدﻗﯾﻘﺔ‬‫إﯾﻘﺎف اﻟﺗﺷﻐﯾل‬‫اﻟﺿﻐط‬‫اﻟﺿﻐط ﻋﻠﻰ زر‬‫ﺳﯾﻧطﻔﺊ‬‫ﻧﻣط اﻹﻋدادات ‪START‬‬‫اﻟﺟﮭﺎز ﻓﻲ أي وﻗت‪.‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﺧﺗﯾﺎر اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟﻠرﺑﺎط‬‫إن اﺳﺗﻌﻣﺎل اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ ﻟرﺑﺎط اﻟذراع ھﺎم ﺟدًا ﻓﻲ دﻗّﺔ ﻗﯾﺎس ﺿﻐط اﻟدم‪ .‬وإذا ﻟم ﯾﻛن ﻣﻘﺎس اﻟرﺑﺎط‬‫ﺻﺣﯾ ًﺣﺎ‪ ،‬ﻓﻘﯾﺎس ﺿﻐط اﻟدم ﺳﯾظﮭر ﻗﯾﻣﺔ ﻏﯾر ﺻﺣﯾﺣﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﻋﺎ ﻋﻠﻰ ﻛل رﺑﺎط‪.‬‬‫ﯾﻛون ﻣﻘﺎس ﻣﺣﯾط اﻟذراع ﻣطﺑو ً‬‫ﯾﺳﺎﻋد ﻛ ﱞل ﻣن اﻟﻣؤﺷر واﻟﻧطﺎق اﻟﻣﻧﺎﺳب‪ ،‬اﻟﻣطﺑوﻋﺎن ﻋﻠﻰ اﻟرﺑﺎط‪ ،‬ﻋﻠﻰ أﻧك ﺗﺳﺗﺧدم اﻟﻣﻘﺎس اﻟﺻﺣﯾﺢ‪.‬‬‫)راﺟﻊ ﺟدول "اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ اﻟرﺑﺎط" ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ(‬‫إذا ﻛﺎن اﻟﻣؤﺷر ﯾﺷﯾر إﻟﻰ ﺧﺎرج اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل اﻟﻣﺣﻠﻲ ﻟﺷراء رﺑﺎط ﺑدﯾل‪.‬‬‫رﺑﺎط اﻟذراع ھو أﺣد اﻟﻣواد اﻟﻣﺳﺗﮭﻠﻛﺔ‪ .‬ﻓﻲ ﺣﺎل أﺻﺑﺢ اﻟرﺑﺎط ﺑﺎﻟﯾﺎ ً ﻓﺎﺳْﺗﺑدﻟﮫ ﻓورا ً ﺑﺂﺧر ﺟدﯾد‪.‬‬‫ﺣﺟم اﻟرﺑﺎط اﻟﻣوﺻﻰ ﺑﮫ‬‫ﺣﺟم اﻟذراع‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬‫‪ 31‬ﺳم إﻟﻰ ‪ 45‬ﺳم‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬‫‪ 22‬ﺳم إﻟﻰ ‪ 42‬ﺳم‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪ 22‬ﺳم إﻟﻰ ‪ 32‬ﺳم‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر‬‫‪ 16‬ﺳم إﻟﻰ ‪ 24‬ﺳم‬‫ﺣﺟم اﻟذراع‪ :‬اﻟﻣﺣﯾط ﻋﻧد ﻋﺿﻠﺔ اﻟذراع‪.‬‬‫ﻛﯾﻔﯾﺔ وﺿﻊ رﺑﺎط اﻟذراع‬‫‪ .1‬ﻟف اﻟرﺑﺎط ﺣول أﻋﻠﻰ اﻟذراع )اﻟﻌﺿد(‪ ،‬ﺣواﻟﻲ ﻣن‬‫‪ 1‬إﻟﻰ ‪ 2‬ﺳم أﻋﻠﻰ اﻟﺟزء اﻟداﺧﻠﻲ ﻣن اﻟﻣرﻓﻖ‪ ،‬ﻛﻣﺎ‬‫ھو ﻣوﺿﺢ‪.‬‬‫ﺿﻊ اﻟرﺑﺎط ﻣﺑﺎﺷرة ﻋﻠﻰ اﻟﺟﻠد‪ ،‬ﻷن اﻟﻣﻼﺑس ﯾﻣﻛﻧﮭﺎ‬‫أن ﺗﺗﺳﺑب ﻓﻲ اﻟﺣﺻول ﻋﻠﻰ ﻧﺑض ﺿﻌﯾف‪ ،‬وﯾﻧﺗﺞ‬‫ﻋﻧﮫ ﺧطﺄ ﻓﻲ اﻟﻘﯾﺎس‪.‬‬‫ﺿد‪ ،‬ﺑﺳﺑب‬‫‪ .2‬ﯾﻣﻛن أن ﯾﻣﻧﻊ اﻟﺗﺿﯾﯾﻖ اﻟﺷدﯾد ﻋﻠﻰ اﻟﻌ ُ‬‫ﻲ ﻛم اﻟﻘﻣﯾص ﺑﺷدة‪ ،‬اﻟﺣﺻول ﻋﻠﻰ ﻗراءات‬‫ط ّ‬‫دﻗﯾﻘﺔ‪.‬‬‫‪ .3‬ﺗﺄﻛد ﻣن وﺟود ﻧﻘﺎط اﻟﻣؤﺷر‬‫اﻟﻣﻧﺎﺳب‪.‬‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫‪CUF-F-SA‬‬‫ﻲ ﻛم اﻟﻘﻣﯾص ﺑﺷدة‪.‬‬‫ﻻ ﺗﻘم ﺑط ّ‬‫‪ 2-1‬ﺳم‬‫اﻟرﺑﺎط‬‫ﺧرطوم اﻟﮭواء‬‫رﻣز ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫أﺣﻛم ﺗﺛﺑﯾت اﻟﻣﺛﺑت‬‫اﻟﻘﻣﺎﺷﻲ‪.‬‬‫ﺿﻣن اﻟﻧطﺎق‬‫اﻟﻣؤﺷر‬‫اﻟﻌرﺑﯾﺔ ‪11‬‬‫واﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ اﻟرﺑﺎط‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫اﻟرﻣوز‬‫رﻣز ﻣوﺿﻊ اﻟﺷرﯾﺎن‬‫▲‬‫‪REF‬‬‫اﻟﻣؤﺷر‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟرﺑﺎط اﻟﺑﺎﻟﻐﯾن‪.‬‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن‪.‬‬‫ﯾﻛون ﻣطﺑو ً‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﺿﺑط اﻟﻌﻼﻣﺔ ﺑﺣﯾث ﺗﻛون ﻋﻧد‬‫ﺿد أو ﻓﻲ ﻣﺣﺎذاة ﻣﻊ‬‫ﺷرﯾﺎن اﻟﻌ ُ‬‫إﺻﺑﻊ اﻟﺑﻧﺻر ﻋﻧد اﻟﺟزء اﻟداﺧﻠﻲ ﻣن‬‫اﻟذراع‪.‬‬‫اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر ﺑدﻻً ﻣن‬‫ﺗﺟﺎوز اﻟﻧطﺎق اﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن واﻟرﺑﺎط‬‫ُ‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن أو اﻟرﺑﺎط ذي اﻟﻧطﺎق‬‫ذي اﻟﻧطﺎق اﻟﻌرﯾض‪.‬‬‫اﻟﻌرﯾض‪.‬‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟﻠرﺑﺎط ذي اﻟﻧطﻖ اﻟﻌرﯾض‪.‬‬‫ﻋﺎ ﻋﻠﻰ اﻟرﺑﺎط ذي اﻟﻧطﻖ اﻟﻌرﯾض‪.‬‬‫ﯾﻛون ﻣطﺑو ً‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ ﻟرﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪ .‬ﯾﻛون‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‪.‬‬‫ﻣطﺑو ً‬‫ً‬‫ﺿﻣن اﻟﻧطﺎق اﻟﻣطﺑوع ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن واﻟرﺑﺎط اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر ﺑدﻻ ﻣن‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن أو اﻟرﺑﺎط ذي اﻟﻧطﺎق‬‫ذي اﻟﻧطﺎق اﻟﻌرﯾض‪.‬‬‫اﻟﻌرﯾض‪.‬‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب ﺑﺎﻟﻧﺳﺑﺔ إﻟﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر‪.‬‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر‪.‬‬‫ﯾﻛون ﻣطﺑو ً‬‫اﺳﺗﺧدم رﺑﺎط اﻟﺑﺎﻟﻐﯾن ﺑدﻻً ﻣن رﺑﺎط‬‫ﻋﺎ ﻋﻠﻰ رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر ورﺑﺎط‬‫ﯾﻛون ﻣطﺑو ً‬‫اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر أو رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر‪.‬‬‫اﻟﺻﻐﯾر‪.‬‬‫رﻗم اﻟﺗﺷﻐﯾﻠﺔ‬‫اﻟﻧطﺎق اﻟﻣﻧﺎﺳب‬

‫‪L‬‬

‫‪A‬‬

‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬

‫‪L‬‬

‫‪W‬‬

‫‪S‬‬

‫اﻟرﺑﺎط ذو اﻟﻧطﺎق اﻟﻌرﯾض‬

‫‪L‬‬

‫‪A‬‬

‫‪S‬‬

‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬

‫‪A‬‬

‫‪S‬‬

‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر‬‫اﻟﻌرﺑﯾﺔ ‪12‬‬‫اﺳﺗﺧدام ﺟﮭﺎز اﻟﻘﯾﺎس‬‫ﻛﯾﻔﯾﺔ إﺟراء ﻋﻣﻠﯾﺎت ﻗﯾﺎس دﻗﯾﻘﺔ‬‫ﻟﻠﺣﺻول ﻋﻠﻰ ﻗﯾﺎس دﻗﯾﻖ ﺗﻣﺎ ًﻣﺎ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪:‬‬‫أرخ ذراﻋك ﻋﻠﻰ اﻟطﺎوﻟﺔ‪ .‬ﻻ ﺗﺿﻊ ﻗد ًﻣﺎ ﻓوق اﻷﺧرى‪ .‬أﺑﻖ ﻗدﻣﯾك ﻣﺳﺗوﯾﺗﯾن ﻋﻠﻰ‬‫اﺟﻠس ﻣﺳﺗرﯾ ًﺣﺎ ﻋﻠﻰ ﻛرﺳﻲ‪.‬‬‫ِ‬‫اﻷرض واﺟﻌل ظﮭرك ﻣﺳﺗﻘﯾ ًﻣﺎ‪.‬‬‫اﺳﺗرخ ﻟﻣدة ﺗﺗراوح ﺑﯾن ﺧﻣس إﻟﻰ ﻋﺷر دﻗﺎﺋﻖ ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫ِ‬‫اﺟﻌل ﻣﻧﺗﺻف اﻟرﺑﺎط ﻋﻧد اﻟﻣﺳﺗوى ﻧﻔﺳﮫ ﻟﻠﻘﻠب‪.‬‬‫اﺑﻖ ﺛﺎﺑﺗًﺎ وﺣﺎﻓظ ﻋﻠﻰ ھدوءك أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫ﻻ ﺗﻘس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓور اﻻﻧﺗﮭﺎء ﻣن ﻣﻣﺎرﺳﺔ اﻟﺗﻣرﯾﻧﺎت اﻟﺑدﻧﯾﺔ أو اﻻﺳﺗﺣﻣﺎم‪ .‬اﺳﺗرخ ﻟﻣدة ﺗﺗراوح ﺑﯾن‬‫ﻋﺷرﯾن وﺛﻼﺛﯾن دﻗﯾﻘﺔ ﻗﺑل إﺟراء ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪.‬‬‫ﺣﺎول ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ ﻛل ﯾوم‪.‬‬‫اﻟﻘﯾﺎس‬‫أﺛﻧﺎء اﻟﻘﯾﺎس‪ ،‬ﻣن اﻟطﺑﯾﻌﻲ أن ﺗﺷﻌر أن اﻟرﺑﺎط ﻣﺷدود ﻟﻠﻐﺎﯾﺔ‪) .‬ﻻ ﺗﻠﺗﻔت ﻟذﻟك(‪.‬‬‫ﺑﻌد اﻟﻘﯾﺎس‬‫ﺑﻌد اﻟﻘﯾﺎس‪ ،‬اﺿﻐط ﻋﻠﻰ زر ‪ START‬ﻹطﻔﺎء اﻟﺟﮭﺎز‪.‬‬‫اﻧزع اﻟرﺑﺎط وﺳﺟل ﺑﯾﺎﻧﺎﺗك‪ .‬ﯾﻧطﻔﺊ اﻟﺟﮭﺎز ﺗﻠﻘﺎﺋﯾًﺎ ﺑﻌد ُﻣﺿﻲ دﻗﯾﻘﺔ واﺣدة ﻣن ﺣﺎﻟﺔ ﻋدم اﻟﺗﺷﻐﯾل‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪13‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‬‫ﻗﺑل اﻟﻘﯾﺎس‪ ،‬اﻗرأ ﻗﺳم "ﻣﻼﺣظﺎت ﻟﻘﯾﺎس دﻗﯾﻖ" ﻓﻲ اﻟﺻﻔﺣﺔ رﻗم ‪.16‬‬‫اﻟﻘﯾﺎس اﻟﻌﺎدي‬‫‪.1‬‬‫ﺿﻊ اﻟرﺑﺎط ﻋﻠﻰ اﻟذراع )ﯾُﻔﺿل اﻟذراع اﻟﯾﺳرى(‪.‬‬‫اﺟﻠس ھﺎدﺋًﺎ ﻓﻲ أﺛﻧﺎء اﻟﻘﯾﺎس‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ زر ‪START‬‬‫ﯾﺗم ﻋرض ﺟﻣﯾﻊ ﺷراﺋﺢ اﻟﺷﺎﺷﺔ‪.‬‬‫ﯾوﻣض اﻟﺻﻔر )‪ (0‬ﻋﻠﻰ اﻟﺷﺎﺷﺔ ﻟﻔﺗرة وﺟﯾزة‪.‬‬‫ﺗﺗﻐﯾّر اﻟﺷﺎﺷﺔ‪ ،‬ﻛﻣﺎ ھو ﻣوﺿّﺢ ﻓﻲ اﻟﺷﻛل ﻋﻠﻰ اﻟﯾﻣﯾن‪ ،‬ﺣﯾث‬‫ﺗﺑدأ ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس‪ .‬ﯾﺑدأ اﻟرﺑﺎط ﻓﻲ اﻻﻧﺗﻔﺎخ‪ .‬ﻣن اﻟطﺑﯾﻌﻲ أن‬‫ﺗﺷﻌر أن اﻟرﺑﺎط ﻣﺷدود ﻟﻠﻐﺎﯾﺔ‪ .‬ﯾُﻌرض ﻣؤﺷر ﺷرﯾط‬‫اﻟﺿﻐط‪ ،‬ﻋﻠﻰ اﻟﺣﺎﻓﺔ اﻟﯾﺳرى ﻣن اﻟﺷﺎﺷﺔ‪ ،‬ﻓﻲ أﺛﻧﺎء ﻧﻔﺦ‬‫اﻟﮭواء‪.‬‬‫ﻣﻠﺣوظﺔ‪ :‬إذا أردت إﯾﻘﺎف ﻧﻔﺦ اﻟﮭواء ﻓﻲ أي وﻗت‪ ،‬ﻓﺎﺿﻐط‬‫ﻋﻠﻰ زر ‪ START‬ﻣرة أﺧرى‪.‬‬‫ﻋرض اﻟﺻﻔر‬‫ﺑدء ﻧ ْﻔﺦ اﻟﮭواء‬‫‪.3‬‬‫ﻋﻧد اﻛﺗﻣﺎل ﻧﻔﺦ اﻟﮭواء‪ ،‬ﯾﺑدأ اﻻﻧﻛﻣﺎش أوﺗوﻣﺎﺗﯾﻛﯾًﺎ وﯾوﻣض‬‫ﺟﺎر إﺟراء اﻟﻘﯾﺎس‪ .‬ﺑﻣﺟرد‬‫)رﻣز اﻟﻘﻠب(‪،‬‬‫ً‬‫ﻣﺷﯾرا إﻟﻰ أﻧﮫ ٍ‬‫ﺗﺣدﯾد اﻟﻧﺑض‪ ،‬ﯾوﻣض اﻟرﻣز ﻣﻊ ﻛل ﻧﺑﺿﺔ‪.‬‬‫ﺟﺎر ﻗﯾﺎس اﻟﺿﻐط‬‫ٍ‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫‪START‬‬‫اﺿﻐط‬‫ﻛل اﻟﺷﺎﺷﺔ‬‫اﻷﺟزاء اﻟﻣﻌروﺿﺔ‬‫ﻣﻼﺣظﺔ‪ :‬ﻓﻲ ﺣﺎﻟﺔ ﻋدم اﻟﺣﺻول ﻋﻠﻰ ﺿﻐط ﻣﻼﺋم‪ ،‬ﯾﺑدأ اﻟﺟﮭﺎز ﻓﻲ‬‫ﻧ ْﻔﺦ اﻟﮭواء ﻣرة أﺧرى ﺗﻠﻘﺎﺋﯾًﺎ‪.‬‬‫ﻟﺗﺟﻧب إﻋﺎدة ﻧﻔﺦ اﻟﮭواء‪ّ ،‬‬‫اطﻠﻊ ﻋﻠﻰ ﻗﺳم "اﻟﻘﯾﺎس‬‫ﻣﻊ ﺿﻐط ‪ " SET‬ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫‪.4‬‬‫ﻋﻧد اﻻﻧﺗﮭﺎء ﻣن اﻟﻘﯾﺎس‪ ،‬ﺗﻌرض اﻟﺷﺎﺷﺔ اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪،‬‬‫واﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ ،‬وﺳرﻋﺔ اﻟﻧﺑض‪.‬‬‫ﯾﻔرغ اﻟرﺑﺎط اﻟﮭواء اﻟﻣﺗﺑﻘﻲ ﺗﻣﺎ ًﻣﺎ‪.‬‬‫ّ‬‫‪.5‬‬‫اﺿﻐط ﻋﻠﻰ زر ‪ START‬اﻹطﻔﺎء اﻟﺟﮭﺎز‪.‬‬‫ﺑﻌد دﻗﯾﻘﺔ واﺣدة ﻣن ﻋدم اﻻﺳﺗﺧدام‪ ،‬ﺳﯾﻧطﻔﺊ اﻟﺟﮭﺎز‬‫أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬اﺟﻌل اﻟﻣدة اﻟزﻣﻧﯾﺔ اﻟﻔﺎرﻗﺔ ﺑﯾن ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻟﻧﻔس‬‫اﻟﺷﺧص ﺛﻼث دﻗﺎﺋﻖ ﻋﻠﻰ اﻷﻗل‪.‬‬‫إﺟراء اﻟﻘﯾﺎس ﻗﯾد‬‫اﻟﺗﻧﻔﯾذ‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﺳرﻋﺔ اﻟﻧﺑض‬‫ﺗﻔرﯾﻎ اﻟﮭواء اﻟﻣﺗﺑﻘﻲ‬‫ﺗﻠﻘﺎﺋﯾًﺎ‬‫اﻟﻌرﺑﯾﺔ ‪14‬‬‫ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‬‫‪ UA-1020-W‬ﻣﺻﻣم ﻟﻠﺗﻌرف ﻋﻠﻰ اﻟﻧﺑض وﻟﻧﻔﺦ اﻟرﺑﺎط ﺑﺎﻟﮭواء إﻟﻰ ﻣﺳﺗوى اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫إذا ﺗﻛرر إﻋﺎدة ﻧ ْﻔﺦ اﻟﮭواء‪ ،‬ﻓﺎﺳﺗﻌن ﺑﺎﻟطرق اﻟﺗﺎﻟﯾﺔ‪.‬‬‫اﻟﻘﯾﺎس ﻣﻊ ﺿﻐط ‪SET‬‬‫ﻓﻲ أﺛﻧﺎء ﻗﯾﺎس ﺿﻐط اﻟدم‪ ،‬ﻣن اﻟﻣﻣﻛن أن ﺗﺣدث إﻋﺎدة ﻟﻧﻔﺦ اﻟﮭواء‪.‬‬‫ﯾﻣﻛن ﺗﺣدﯾد ﻗﯾﻣﺔ ﺿﻐط ﺛﺎﺑﺗﺔ ﻟﺗﺟﻧب إﻋﺎدة ﻧﻔﺦ اﻟﮭواء‪.‬‬‫‪.1‬‬‫ﻓﻲ اﻟﺧطوة اﻟﺳﺎدﺳﺔ ﻣن ﻋﻣﻠﯾﺔ ﺗﻌدﯾل اﻟﺳﺎﻋﺔ )اﻧظر اﻟﺻﻔﺣﺔ رﻗم‬‫‪ ،(10‬اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﻠذھﺎب إﻟﻰ إﻋدادات ﻧﻣط‬‫اﻟﺿﻐط‪ .‬اﻹﻋدادات اﻟﺣﺎﻟﯾﺔ ﺗوﻣض‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر أو ﻻﺧﺗﯾﺎر ﻗﯾﻣﺔ ﺿﻐط ﺣواﻟﻲ‬‫‪ mmHg 30‬أﻋﻠﻰ ﻣن ﺿﻐطك اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣﺗوﻗﻊ ﻣن‬‫ﺑﯾن ﻣﺎ ﯾﻠﻲ‪.‬‬‫أو‬‫‪ : AUTO‬ﺗطﺑﯾﻖ اﻟﺿﻐط اﻷوﺗوﻣﺎﺗﯾﻛﻲ )اﻟﻘﯾﻣﺔ اﻻﻓﺗراﺿﯾﺔ(‬‫‪ : 180‬ﻗﯾﻣﺔ ﺿﻐط ﻣﻘدارھﺎ ‪) mmHg 180‬ﺛﺎﺑﺗﺔ(‬‫‪ : 210‬ﻗﯾﻣﺔ ﺿﻐط ﻣﻘدارھﺎ ‪) mmHg 210‬ﺛﺎﺑﺗﺔ(‬‫‪ : 240‬ﻗﯾﻣﺔ ﺿﻐط ﻣﻘدارھﺎ ‪) mmHg 240‬ﺛﺎﺑﺗﺔ(‬‫‪.3‬‬‫‪ TriCheck‬إﯾﻘﺎف اﻟﺗﺷﻐﯾل‬‫ﻧﻣط اﻹﻋدادات‬‫اﺿﻐط ﻋﻠﻰ زر ‪ SET‬ﻟﻠذھﺎب إﻟﻰ ﻧﻣط إﻋدادات ‪.TriCheck‬‬‫اﺿﻐط ﻋﻠﻰ زر ‪ START‬ﻹطﻔﺎء اﻟﺟﮭﺎز‪ .‬ﺑﻌد ﺛﻼث دﻗﺎﺋﻖ ﻣن ﻋدم اﻻﺳﺗﺧدام‪ ،‬ﺳﯾﻧطﻔﺊ اﻟﺟﮭﺎز‬‫أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫ﺳﯾُﺟرى اﻟﻘﯾﺎس اﻟﺗﺎﻟﻲ ﻣﻊ ﻗﯾﻣﺔ اﻟﺿﻐط اﻟﺟدﯾدة‪.‬‬‫إﺟراء اﻟﻘﯾﺎس ﻣﻊ اﺧﺗﯾﺎر اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣرﻏوب ﻓﯾﮫ‬‫اﺳﺗﺧدم ھذه اﻟطرﯾﻘﺔ ﻋﻧدﻣﺎ ﺗﺣدث إﻋﺎدة ﻧﻔﺦ اﻟﮭواء ﺑﺷﻛل ﻣﺗﻛرر ﺣﺗﻰ إذا ﻛﺎﻧت ﻗﯾﻣﺔ‬‫اﻟﺿﻐط ﻣﺿﺑوطﺔ ﻋﻠﻰ ‪ 240‬ﻓﻲ اﻟﻌﻣﻠﯾﺔ أﻋﻼه أو ﻋﻧدﻣﺎ ﻻ ﺗُﻌرض اﻟﻧﺗﺎﺋﺞ ﺣﺗﻰ إذا ﻗل‬‫اﻟﺿﻐط إﻟﻰ ‪ mmHg 20‬أو أﻗل‪.‬‬‫‪.1‬‬‫ﺿﻊ اﻟرﺑﺎط ﻋﻠﻰ اﻟذراع )ﯾُﻔﺿل اﻟذراع اﻟﯾﺳرى(‪.‬‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫‪.2‬‬‫اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ زر ‪ START‬ﺣﺗﻰ ﯾظﮭر‬‫رﻗم أﻋﻠﻰ ﻣن ﺿﻐطك اﻻﻧﻘﺑﺎﺿﻲ اﻟﻣﺗوﻗﻊ‬‫ﺑـ ‪ 30‬إﻟﻰ ‪.mmHg 40‬‬‫اﺿﻐط ﺑﺎﺳﺗﻣرار ﻋﻠﻰ اﻟزر‬‫‪.3‬‬‫ﻋﻧد اﻟوﺻول إﻟﻰ اﻟرﻗم اﻟﻣطﻠوب‪ ،‬اﺗرك زر ‪START‬‬‫ﻟﺑدء اﻟﻘﯾﺎس‪ .‬ﺗﺎﺑﻊ إﺟراء ﻋﻣﻠﯾﺔ ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط‬‫اﻟدم ﻛﻣﺎ ھو ﻣوﺿﺢ ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺳﺎﺑﻘﺔ‪.‬‬‫‪START‬‬‫ﺣرر اﻟزر ﻋﻧد ﻣﺳﺗوى‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﻣطﻠوب‬‫راﺟﻊ اﻟﺻﻔﺣﺔ اﻟﺳﺎﺑﻘﺔ ﻟﻼطﻼع ﻋﻠﻰ‬‫إﺟراء اﻟﻘﯾﺎس‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪15‬‬‫ﻣﻼﺣظﺎت ﻟﻘﯾﺎس دﻗﯾﻖ‬‫اﺟﻠس ﻓﻲ وﺿﻊ ﻣرﯾﺢ‪ .‬ﺿﻊ ذراﻋك ﻋﻠﻰ طﺎوﻟﺔ ﺑﺣﯾث ﺗﻛون راﺣﺔ ﯾدك ﻣواﺟﮭﺔ ﻷﻋﻠﻰ واﻟرﺑﺎط ﻋﻧد‬‫ﻣﺳﺗوى اﻟﻘﻠب‪.‬‬‫اﺳﺗرخ ﻟﻣدة ﺣواﻟﻲ ﻣن ﺧﻣس إﻟﻰ ﻋﺷر دﻗﺎﺋﻖ ﻗﺑل إﺟراء اﻟﻘﯾﺎس‪ .‬إذا ﻛﺎﻧت ﺗﻌﺗرﯾك ﻓرﺣﺔ ﻋﺎرﻣﺔ أو ﺗﻣر‬‫ﺑﺣﺎﻟﺔ اﻛﺗﺋﺎب ﻧﺗﯾﺟﺔ ﻟﻠﺿﻐوطﺎت اﻟﻌﺎطﻔﯾﺔ‪ ،‬ﻓﺳﯾﻧﻌﻛس ذﻟك ﻣﻧﺗ ًﺟﺎ ﻗراءة أﻋﻠﻰ )أو أﻗل( ﻣن اﻟﻘراءة اﻟطﺑﯾﻌﯾﺔ‬‫ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬وﺳﺗﻛون ﻗراءة ﺳرﻋﺔ اﻟﻧﺑض أﺳرع ﻣن اﻟﻣﻌدل اﻟطﺑﯾﻌﻲ‪.‬‬‫ﯾﺗﺑﺎﯾن ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟﻠﻔرد ﺑﺎﺳﺗﻣرار ﺗﺑﻌًﺎ ﻟﻠﻌﻣل اﻟذي ﺗﻘوم ﺑﮫ واﻟطﻌﺎم اﻟذي ﺗﺗﻧﺎوﻟﮫ‪ .‬ﯾﻣﻛن أن ﯾﻛون ﻟﻣﺎ‬‫ﺗﺷرﺑﮫ ﺗﺄﺛﯾر ﻗوي وﺳرﯾﻊ ﻋﻠﻰ ﺿﻐط دﻣك‪.‬‬‫ﯾﻌﺗﻣد ھذا اﻟﺟﮭﺎز ﻓﻲ ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس اﻟﺗﻲ ﯾُﺟرﯾﮭﺎ ﻋﻠﻰ ﺿرﺑﺎت اﻟﻘﻠب‪ .‬ﻓﺈذا ﻛﺎن ﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟدﯾك‬‫ﺿﻌﯾﻔًﺎ ﺟدًا أو ﻏﯾر ﻣﻧﺗظم‪ ،‬ﻓﻘد ﯾﺟد اﻟﺟﮭﺎز ﺻﻌوﺑﺔ ﻓﻲ ﺗﺣدﯾد ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪.‬‬‫إذا اﻛﺗﺷف اﻟﺟﮭﺎز ﺣﺎﻟﺔ ﻏﯾر طﺑﯾﻌﯾﺔ‪ ،‬ﻓﺳﯾوﻗف ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس وﯾُظﮭر رﻣز اﻟﺧطﺄ‪ .‬راﺟﻊ اﻟﺻﻔﺣﺔ رﻗم ‪6‬‬‫ﻟوﺻف اﻟرﻣوز‪.‬‬‫ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ُﻣﺧﺻص ﻟﻼﺳﺗﺧدام ﻣن ﻗِﺑل اﻟﺑﺎﻟﻐﯾن‪ .‬اﺳﺗﺷر طﺑﯾﺑك ﻗﺑل اﺳﺗﺧدام ھذا‬‫اﻟﺟﮭﺎز ﻣﻊ أﺣد اﻷطﻔﺎل‪ .‬ﯾﺟب ﻋدم اﺳﺗﺧدام اﻷطﻔﺎل اﻟﺟﮭﺎز دون إﺷراف‪.‬‬‫ﯾﻣﻛن أن ﯾﺗﺄﺛر أداء ﺟﮭﺎز ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﺗﻠﻘﺎﺋﻲ ﺑﺳﺑب اﻟزﯾﺎدة اﻟﺑﺎﻟﻐﺔ ﻓﻲ درﺟﺔ اﻟﺣرارة أو‬‫اﻟرطوﺑﺔ أو اﻻرﺗﻔﺎع ﻋن ﺳطﺢ اﻟﺑﺣر‪.‬‬‫ﻗﯾﺎس ‪TriCheckTM‬‬‫ﻧﻣط ‪TriCheckTM‬ﯾﺄﺧذ ﺛﻼﺛﺔ ﻗﯾﺎﺳﺎت ﻣﺗﺗﺎﻟﯾﺔ أوﺗوﻣﺎﺗﯾﻛﯾًﺎ وﺗﻌرض اﻟﻘﯾم اﻟﻣﺗوﺳطﺔ ﻟﻠﺛﻼﺛﺔ ﻗﯾﺎﺳﺎت‪.‬‬‫اﺧﺗﯾﺎر ﻧﻣط ‪TriCheckTM‬‬‫‪.1‬‬‫ﻓﻲ إﻋدادات ﻧﻣط اﻟﺿﻐط‪ ،‬اﺿﻐط ﻋﻠﻰ زر ‪SET‬‬‫ﻟدﺧول ﻧﻣط إﻋدادات ‪ TriCheckTM‬اﻹﻋدادات‬‫اﻟﺣﺎﻟﯾﺔ ﺗوﻣض‪.‬‬‫‪.2‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر‬‫‪ TriCheckTM‬أو إﻏﻼﻗﺔ‪.‬‬‫ﻓﺗﺢ‪ :‬ﻧﻣط ‪TriCheckTM‬‬‫إﻏﻼق‪ :‬ﻧﻣط اﻟﻘﯾﺎس اﻟﻌﺎدي )اﻟﻘﯾﻣﺔ اﻻﻓﺗراﺿﯾﺔ(‬‫ﻧﻣط‬‫‪.3‬‬‫اﺿﻐط ﻋﻠﻰ اﻟزر ‪ START‬أو ‪ SET‬ﻹطﻔﺎء‬‫اﻟﺟﮭﺎز‪ .‬ﺑﻌد ﺛﻼث دﻗﺎﺋﻖ ﻣن ﻋدم اﻻﺳﺗﺧدام‪ ،‬ﺳﯾﻧطﻔﺊ‬‫اﻟﺟﮭﺎز أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫أو‬‫ﻟﻔﺗﺢ‬‫ﻓﻲ إﻋدادات ﻧﻣط اﻟﺿﻐط‪،‬‬‫اﺿﻐط‬‫اﻹﻋدادات اﻟﺣﺎﻟﯾﺔ ﺗوﻣض‬‫أو‬‫ﺗﻐﯾﯾر اﻹﻋدادات‬‫‪START‬‬‫اﻟﻌرﺑﯾﺔ ‪16‬‬‫أو‬‫اﻟﻘﯾﺎس ﺑﺎﺳﺗﺧدام ﻧﻣط ‪TriCheckTM‬‬‫‪.1‬‬‫اﺿﻐط ﻋﻠﻰ زر ‪. START‬‬‫ﯾﺗم ﻋرض ﺟﻣﯾﻊ ﺷراﺋﺢ اﻟﺷﺎﺷﺔ‪.‬‬‫ﯾﻌرض اﻟرﻗم ﺻﻔر )‪ (0‬وھو ﯾوﻣض ﻟﻔﺗرة ﻗﺻﯾرة‬‫وﯾﺑدأ اﻟﻘﯾﺎس اﻷول‪.‬‬‫‪.2‬‬‫ﻋﻧد اﻻﻧﺗﮭﺎء ﻣن اﻟﻘﯾﺎس‪ ،‬ﺗﻌرض اﻟﺷﺎﺷﺔ اﻟﺿﻐط‬‫اﻻﻧﻘﺑﺎﺿﻲ‪ ،‬واﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ ،‬وﺳرﻋﺔ اﻟﻧﺑض‪ ،‬ﺛم‬‫ﯾﺑدأ ﻋد ﺗﻧﺎزﻟﻲ ﻟﻣدة دﻗﯾﻘﺔ ﻹﺟراء اﻟﻘﯾﺎس اﻟﺛﺎﻧﻲ‪.‬‬‫‪.3‬‬‫ﺑﻌد دﻗﯾﻘﺔ واﺣدة‪ ،‬ﯾﺑدأ اﻟﻘﯾﺎس اﻟﺛﺎﻧﻲ‪.‬‬‫‪.4‬‬‫ﻋﻧد اﻻﻧﺗﮭﺎء ﻣن اﻟﻘﯾﺎس‪ ،‬ﯾظﮭر ﻣﺗوﺳط اﻟﻘراءﺗﯾن‬‫اﻷوﻟﻰ واﻟﺛﺎﻧﯾﺔ‪ ،‬ﺛم ﯾﺑدأ ﻋد ﺗﻧﺎزﻟﻲ ﻟﻣدة دﻗﯾﻘﺔ ﻹﺟراء‬‫اﻟﻘﯾﺎس اﻟﺛﺎﻟث‪.‬‬‫‪.5‬‬‫ﺑﻌد دﻗﯾﻘﺔ واﺣدة‪ ،‬ﯾﺑدأ اﻟﻘﯾﺎس اﻟﺛﺎﻟث‪.‬‬‫‪.6‬‬‫ﻋﻧد اﻻﻧﺗﮭﺎء ﻣن اﻟﻘﯾﺎس‪ ،‬ﯾُﻌرض ﻣﺗوﺳط ﻗراءات‬‫اﻟﺛﻼث وﯾ ّ‬‫ُﺧزن ﻓﻲ اﻟذاﻛرة‪.‬‬‫ﻣﻼﺣظﺎت‪:‬‬‫ﻓﻲ أﺛﻧﺎء اﻟﻘﯾﺎس‪ ،‬ﯾُﻌرض رﻣز ‪TriCheck‬‬‫ﻹﻟﻐﺎء اﻟﻘﯾﺎس‪ ،‬اﺿﻐط زر ‪ . START‬ﻓﻲ ھذه‬‫اﻟﺣﺎﻟﺔ‪ ،‬ﻻ ﺗﺧزن أي ﺑﯾﺎﻧﺎت ﻓﻲ اﻟذاﻛرة‪.‬‬‫ﻋﻧدﻣﺎ ﯾظﮭر رﻣز ﺧطﺄ ﻓﻲ ﺗﺛﺑﯾت اﻟرﺑﺎط ﻓﻲ أﺛﻧﺎء‬‫اﻟﻘﯾﺎس اﻷول‪ ،‬أﻟﻎ اﻟﻘﯾﺎس‪ ،‬وأﻋد ﺗﺛﺑﯾت اﻟرﺑﺎط‬‫ﺳﺎ ﺟدﯾدًا‪.‬‬‫ﺑﺷﻛل ﺻﺣﯾﺢ واﺑدأ ﻗﯾﺎ ً‬‫ﺑﻌد اﻟﻘﯾﺎس‪ ،‬ﯾ ّ‬‫ُﺧزن ﻣﺗوﺳط ﺑﯾﺎﻧﺎت اﻟﻘﯾﺎﺳﺎت اﻟﺛﻼﺛﺔ‬‫ﻓﻲ اﻟذاﻛرة‪.‬‬‫ﻟن ﺗ ُ ّ‬‫ﺧزن أي ﺑﯾﺎﻧﺎت‪ ،‬ﻋﻧد إﻟﻐﺎء اﻟﻘﯾﺎﺳﺎت‪ ،‬وﻗﺑل‬‫اﻛﺗﻣﺎل اﻟﻘﯾﺎس اﻟﺛﺎﻟث‪.‬‬‫‪TM‬‬‫‪.‬‬‫ﻋﻧد ﻣﺳﺗوى اﻟﻘﻠب‬‫اﺿﻐط‬‫ﻛل اﻟﺷﺎﺷﺔ‬‫اﻷﺟزاء اﻟﻣﻌروﺿﺔ‬‫اﻟﻘﯾﺎس اﻷول‬‫ﻧﺗﺎﺋﺞ‬‫اﻟﻘﯾﺎس اﻷول‬‫ﺑﻌد دﻗﯾﻘﺔ‬‫اﻟﻘﯾﺎس اﻟﺛﺎﻧﻲ‬‫ﻣﺗوﺳط اﻟﻘﯾﺎس اﻷول‬‫واﻟﻘﯾﺎس اﻟﺛﺎﻧﻲ‬‫اﻟﻘﯾﺎس اﻟﺛﺎﻟث‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫ﻟﻠﻘﯾﺎﺳﺎت اﻟﺛﻼﺛﺔ‬‫اﻟﻌرﺑﯾﺔ ‪17‬‬‫‪START‬‬‫ﺑﻌد دﻗﯾﻘﺔ‬‫اﺳﺗدﻋﺎء اﻟﺑﯾﺎﻧﺎت ﻣن اﻟذاﻛرة‬‫ﻣﻠﺣوظﺔ‪ :‬ﯾﻣﻛن ﻟﮭذا ﻟﺟﮭﺎز ﺗﺳﺟﯾل آﺧر ﺗﺳﻌﯾن ﻣﺟﻣوﻋﺔ ﻣن اﻟﺑﯾﺎﻧﺎت ﻓﻲ اﻟذاﻛرة أوﺗوﻣﺎﺗﯾﻛﯾًﺎ‪.‬‬‫‪ .1‬اﺿﻐط ﻋﻠﻰ اﻟزر أو ‪.‬‬
‫ﯾُﻌرض ﻣﺗوﺳط ﻛل اﻟﻘراءات وﻋدد اﻟﺑﯾﺎﻧﺎت‪.‬‬‫)إذا ﻟم ﯾﻛن ھﻧﺎك ﺑﯾﺎﻧﺎت ﯾُﻌرض "‪ ."0‬اﺿﻐط ﻋﻠﻰ زر‬‫‪ ،‬أو ‪ START‬ﻹطﻔﺎء اﻟﺟﮭﺎز‪(.‬‬‫‪ .2‬ﻓﻲ ﻛل ﻣرة ﯾﺗم ﻓﯾﮭﺎ اﻟﺿﻐط ﻋﻠﻰ اﻟزر )أو اﻟزر‬‫ﻟﻌرض اﻟﺑﯾﺎﻧﺎت ﺑﺎﻟﺗرﺗﯾب اﻟﻌﻛﺳﻲ(‪ ،‬ﺗُﻌرض‬‫ﺑﯾﺎﻧﺎت اﻟذاﻛرة ﻛﺎﻟﺗﺎﻟﻲ‪.‬‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻛل اﻟﻘﯾﺎﺳﺎت اﻟﺻﺑﺎﺣﯾﺔ اﻟﻣﺄﺧوذة‬‫ﺑﯾن اﻟﺳﺎﻋﺔ ‪ 4:00‬و‪.9:59‬‬‫)ﻋﻠﻰ ﺳﺑﯾل اﻟﻣﺛﺎل‪ 10 ،‬ﻗﯾﺎﺳﺎت‪ .‬إذا ﻟم ﺗﻛن ھﻧﺎك‬‫ﺑﯾﺎﻧﺎت ﯾُﻌرض "‪(."--‬‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻛل اﻟﻘﯾﺎﺳﺎت اﻟﻣﺳﺎﺋﯾﺔ اﻟﻣﺄﺧوذة‬‫ﺑﯾن اﻟﺳﺎﻋﺔ ‪ 18:00‬و‪.1:59‬‬‫)ﻋﻠﻰ ﺳﺑﯾل اﻟﻣﺛﺎل‪ 9 ،‬ﻗﯾﺎﺳﺎت‪ .‬إذا ﻟم ﺗﻛن ھﻧﺎك‬‫ﺑﯾﺎﻧﺎت ﯾُﻌرض "‪(."--‬‬‫أﺣدث اﻟﺑﯾﺎﻧﺎت )اﻟﻌدد‪ ،‬ﻋﻠﻰ ﺳﺑﯾل اﻟﻣﺛﺎل‪ ،‬رﻗم ‪(35‬‬‫ﺛوان ﻣن ﻋرض رﻗم اﻟﺑﯾﺎﻧﺎت‪ ،‬ﺗُﻌرض‬‫ﺑﻌد ﺛﻼث‬‫ٍ‬‫ﺑﯾﺎﻧﺎت اﻟﻘﯾﺎس‪.‬‬‫اﺿﻐط ﻋﻠﻰ‬‫أو‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﻧﺑض‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫ﻟﻛل اﻟﺑﯾﺎﻧﺎت‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﻧﺑض‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫ﻣن اﻟﺑﯾﺎﻧﺎت اﻟﺻﺑﺎﺣﯾﺔ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﺿﻐط اﻻﻧﺑﺳﺎطﻲ‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط ﻟﻠﻧﺑض‬‫اﻟﻣﻌدل اﻟﻣﺗوﺳط‬‫ﻣن اﻟﺑﯾﺎﻧﺎت اﻟﻣﺳﺎﺋﯾﺔ‬‫آﺧر اﻟﺑﯾﺎﻧﺎت )رﻗم ‪(1‬‬‫ﺛوان ﻣن ﻋرض رﻗم اﻟﺑﯾﺎﻧﺎت‪ ،‬ﺗُﻌرض ﺑﯾﺎﻧﺎت‬‫ﺑﻌد ﺛﻼث‬‫ٍ‬‫اﻟﻘﯾﺎس‪.‬‬‫‪ .3‬ﺑﻌد ﻋرض آﺧر اﻟﺑﯾﺎﻧﺎت‪ ،‬اﺿﻐط ﻋﻠﻰ زر‬‫ﻟﻠﻣﻌدل اﻟﻣﺗوﺳط ﻟﻛل اﻟﻘﯾﺎﺳﺎت‪.‬‬‫ﻟﻠﻌودة‬‫اﻧﻘﺑﺎﺿﻲ‬‫اﻧﺑﺳﺎطﻲ‬‫اﻟﻧﺑض‬‫أﺣدث اﻟﺑﯾﺎﻧﺎت‬‫‪ .4‬اﺿﻐط ﻋﻠﻰ زر ‪ START‬ﻹطﻔﺎء اﻟﺟﮭﺎز‪ .‬ﯾﻧطﻔﺊ‬‫اﻟﺟﮭﺎز ﺗﻠﻘﺎﺋﯾًﺎ ﺑﻌد ُﻣﺿﻲ دﻗﯾﻘﺔ واﺣدة ﻣن ﺣﺎﻟﺔ ﻋدم‬‫اﻟﺗﺷﻐﯾل‪.‬‬‫اﻧﻘﺑﺎﺿﻲ‬‫اﻧﺑﺳﺎطﻲ‬‫اﻟﻧﺑض‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻘدﯾﻣﺔ )اﻷﺧﯾرة(‬‫اﻟﻌرﺑﯾﺔ ‪18‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣؤﺷر ‪IHB/AFib‬؟‬‫ﻋﻧد اﻛﺗﺷﺎف ﺟﮭﺎز اﻟﻘﯾﺎس وﺟود ْ‬‫ﻧظم ﻏﯾر ﻣﻧﺗظم أﺛﻧﺎء ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس‪ ،‬ﺳﯾظﮭر اﻟﻣؤﺷر ‪ IHB/AFib‬ﻋﻠﻰ‬‫اﻟﺷﺎﺷﺔ ﻣﻊ ﻗﯾم اﻟﻘﯾﺎس‪.‬‬‫ﻣﻠﺣوظﺔ‪ :‬ﻧُوﺻﻲ ﺑﺎﻟﺗواﺻل ﻣﻊ طﺑﯾﺑك إذا رأﯾت ﻣؤﺷر‬‫‪ IHB/AFib‬ﺑﺷﻛل ﻣﺗﻛرر‪.‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑـ ‪AFib‬؟‬‫ﻧظرا إﻟﻰ اﻹﺷﺎرات اﻟﻛﮭرﺑﺎﺋﯾﺔ اﻟواﻗﻌﺔ ﻓﻲ اﻟﻘﻠب وﯾﻧﺷر اﻟدم ﻓﻲ ﺟﻣﯾﻊ أﻧﺣﺎء اﻟﺟﺳم‪ .‬ﯾﺣدث‬‫ﯾﻧﻘﺑض اﻟﻘﻠب ً‬‫ُ‬‫ُ‬‫اﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪ (AFib‬ﻋﻧدﻣﺎ ﺗﺻﺑﺢ اﻹﺷﺎرة اﻟﻛﮭرﺑﺎﺋﯾﺔ ﻓﻲ اﻷذﯾن ﻣﺷوﺷﺔ‪ ،‬ﻣﺎ ﯾؤدي إﻟﻰ اﺿطراﺑﺎت‬‫ﻓﻲ اﻟﻔواﺻل اﻟزﻣﻧﯾﺔ ﺑﯾن اﻟﺿرﺑﺎت‪ .‬وﯾﻣﻛن أن ﯾﺳﺑب اﻟرﺟﻔﺎن اﻷُذﯾﻧﻲ )‪ (AFib‬رﻛود اﻟدم ﻓﻲ اﻟﻘﻠب‪ ،‬ﻣﺎ‬‫ﻟﺗﻌرض ﻟﻧوﺑﺔ ﻗﻠﺑﯾﺔ‪.‬‬‫ﯾﺷ ّﻛل ﺑﺳﮭوﻟﺔ ﺟﻠطﺎت دﻣوﯾﺔ ﺗﺗﺳﺑب ﻻﺣﻘًﺎ ﻓﻲ ا ّ‬‫ﻣؤﺷر ﺷرﯾط اﻟﺿﻐط‬‫ﯾراﻗب اﻟﻣؤﺷر ﻣدى اﻟﺗﻘدّم ﻓﻲ ﻋﻣﻠﯾﺔ ﻗﯾﺎس اﻟﺿﻐط‪.‬‬‫ﺟﺎر ﺗﻔرﯾﻎ اﻟﮭواء‬‫ٍ‬‫ﺟﺎر ﻗﯾﺎس اﻟﺿﻐط‬‫ٍ‬‫ﻧ ْﻔﺦ اﻟﮭواء ﻗﯾد اﻟﺗﻧﻔﯾذ‬‫ﻋﻣﻠﯾﺔ اﻟﻘﯾﺎس ﻗﯾد اﻟﺗﻧﻔﯾذ‬‫اﻛﺗﻣل ﻧ ْﻔﺦ اﻟﮭواء‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﯾﺷﯾر ﻛل ﺟزء ﻓﻲ ﺷرﯾط اﻟﻣؤﺷر إﻟﻰ ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ‬‫اﻟﻌﺎﻟﻣﯾﺔ ﻟﺿﻐط اﻟدم اﻟﻣوﺻوف ﻓﻲ اﻟﺻﻔﺣﺔ اﻟﺗﺎﻟﯾﺔ‪.‬‬‫ﻣؤﺷر ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﺷدﯾد‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﻣﺗوﺳط‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﺧﻔﯾف‬‫ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫طﺑﯾﻌﻲ‬‫ﻣﺛﺎﻟﻲ‬‫ﻣﺛﺎل‪:‬‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﻣﺗوﺳط‬‫ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ اﻟﺧﻔﯾف‬‫‪ :‬ﯾﻌرض اﻟﻣؤﺷر ﺷرﯾﺣﺔ ﺗﺑﻌًﺎ ﻟﻠﺑﯾﺎﻧﺎت اﻟﺣﺎﻟﯾﺔ وﺑﻣﺎ ﯾﺗواﻓﻖ‬‫ﻣﻊ ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ‪.‬‬‫ﻣرﺗﻔﻊ إﻟﻰ طﺑﯾﻌﻲ‬‫اﻟﻌرﺑﯾﺔ ‪19‬‬‫ﻧُﺑذة ﻋن ﻣﺳﺗوى ﺿﻐط اﻟدم‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﻣﺳﺗوى ﺿﻐط اﻟدم؟‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم ھو ﻗوة دﻓﻊ اﻟدم ﻟﺟدران اﻟﺷراﯾﯾن‪ .‬وﻋﻧدﻣﺎ ﯾﻧﻘﺑض اﻟدم‪ ،‬ﯾﺣدث اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪.‬‬‫وﻋﻧدﻣﺎ ﯾﻧﺑﺳط اﻟدم‪ ،‬ﯾﺣدث اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪ .‬وﯾﻛون اﻟﻣﻠﻠﯾﻣﺗر اﻟزﺋﺑﻘﻲ )‪ (mmHg‬وﺣدة ﻗﯾﺎس ﻣﺳﺗوى‬‫ﺿﻐط اﻟدم‪ .‬وﯾُﻣﺛل اﻟﺿﻐط اﻷﺳﺎﺳﻲ ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟطﺑﯾﻌﻲ ﻟﻠﻔرد‪ ،‬وھو ّأول ﻣﺎ ﯾُﻘﺎس ﻓﻲ اﻟﺻﺑﺎح‬‫ﺣﯾث ﯾﻛون اﻟﻔرد ﻣﺳﺗرﺧﯾًﺎ وﻗﺑل ﺗﻧﺎوﻟﮫ اﻟطﻌﺎم‪.‬‬‫ﻣﺎ اﻟﻣﻘﺻود ﺑﺎرﺗﻔﺎع ﺿﻐط اﻟدم وﻛﯾف ﯾﻣﻛن اﻟﺗﺣﻛم ﻓﯾﮫ؟‬‫ارﺗﻔﺎع ﺿﻐط اﻟدم ھو ﺣﺎﻟﺔ ﻣن ﺿﻐط اﻟدم اﻟﺷرﯾﺎﻧﻲ اﻟﻣرﺗﻔﻊ ﻏﯾر اﻟطﺑﯾﻌﻲ‪ ،‬اﻟذي ﯾﻣﻛن أن ﯾﺗﺳﺑب‪ ،‬ﻓﻲ ﺣﺎﻟﺔ‬‫اﻟﺗﻌرض ﻟﻣﺷﻛﻼت ﺻﺣﯾﺔ‪ ،‬ﻣن ﺑﯾﻧﮭﺎ اﻟﻧوﺑﺔ اﻟﻘﻠﺑﯾﺔ‪ .‬وﯾﻣﻛن اﻟﺗﺣﻛم ﻓﻲ ارﺗﻔﺎع ﺿﻐط‬‫ﻋدم اﻟﺗﺣﻛم ﻓﯾﮫ‪ ،‬ﻓﻲ‬‫ّ‬‫اﻟدم ﻋن طرﯾﻖ ﺗﻐﯾﯾر ﻧﻣط اﻟﺣﯾﺎة‪ ،‬وﺗﻔﺎدي ﺣﺎﻻت اﻹﺟﮭﺎد‪ ،‬وﺑﺗﻧﺎول اﻟﻌﻼج ﺗﺣت اﻹﺷراف اﻟطﺑﻲ‪.‬‬‫ﻟﻠوﻗﺎﯾﺔ ﻣن ارﺗﻔﺎع ﺿﻐط اﻟدم أو اﻟﺗﺣﻛم ﻓﯾﮫ‪:‬‬‫ﻣﺎرس اﻟرﯾﺎﺿﺔ ﺑﺎﻧﺗظﺎم‬‫ﺗﺟﻧب اﻟﺗدﺧﯾن‬‫اﺧﺿﻊ ﻟﻔﺣوﺻﺎت طﺑﯾﺔ ﻣﻧﺗظﻣﺔ‬‫ﻗﻠّل ﻣن ﺗﻧﺎول اﻷﻣﻼح واﻟدھون‬‫ﺣﺎﻓظ ﻋﻠﻰ وزﻧك اﻟﻣﺛﺎﻟﻲ‬‫ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟‬‫إن ﻗﯾﺎس ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ ﻋﯾﺎدة أو ﻋﻧد اﻟطﺑﯾب ﯾﻣﻛن أن ﯾﺛﯾر اﻟﻣﺧﺎوف وﯾُﻌطﻲ ﻗراءة ﻣرﺗﻔﻌﺔ‪،‬‬‫ﺑﻣﻘدار ‪ 25‬إﻟﻰ ‪ mmHg 30‬أﻋﻠﻰ ﻣن اﻟﻘﯾﺎس ﻓﻲ اﻟﻣﻧزل‪ .‬ﯾﻘﻠل اﻟﻘﯾﺎس ﻓﻲ اﻟﻣﻧزل ﻣن اﻟﺗﺄﺛﯾرات اﻟﺧﺎرﺟﯾﺔ‬‫ﻋﻠﻰ ﻗراءات ﻣﺳﺗوى ﺿﻐط اﻟدم‪ ،‬وﯾﻣﺛل إﺿﺎﻓﺔ إﻟﻰ ﻗراءات اﻟطﺑﯾب‪ ،‬وﯾﻘدم ﺳﺟﻼً ﻣن اﻟﻘراءات اﻟﻛﺎﻣﻠﺔ‬‫واﻷﻛﺛر دﻗﺔ ﺑﺎﻟﻧﺳﺑﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫ﺗﺻﻧﯾف ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‬‫وﺿﻌت ﻣﻧظﻣﺔ اﻟﺻﺣﺔ اﻟﻌﺎﻟﻣﯾﺔ ﻣﻌﺎﯾﯾر‬‫ﻟﺗﻘﯾﯾم ﻣﺳﺗوى ﺿﻐط اﻟدم اﻟﻣرﺗﻔﻊ‪ ،‬دون‬‫اﻟﻧظر إﻟﻰ اﻟﻌﻣر‪ ،‬ﻛﻣﺎ ﯾﺗﺿﺢ ذﻟك ﻓﻲ‬‫اﻟرﺳم اﻟﺗﺧطﯾطﻲ‪.‬‬‫اﻻﺧﺗﻼﻓﺎت ﻓﻲ ﺿﻐط اﻟدم‬‫ﯾﺧﺗﻠف ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟﻠﻔرد‬‫ﻛﺑﯾرا ﻋﻠﻰ ﻣدار اﻟﯾوم‬‫اﺧﺗﻼﻓًﺎ‬‫ً‬‫واﻟﻣوﺳم‪ .‬وﯾﻣﻛن أن ﯾﺗراوح‬‫ﻣﻘدار اﻻﺧﺗﻼف ﺑﯾن ‪ 30‬و‪mmHg 50‬‬‫ﻧظرا إﻟﻰ اﻟظروف اﻟﻣﺧﺗﻠﻔﺔ طوال اﻟﯾوم‪ .‬وﯾﻣﻛن أن ﺗﻛون اﻻﺧﺗﻼﻓﺎت ﻟدى اﻷﻓراد اﻟذﯾن ﯾﻌﺎﻧون ارﺗﻔﺎع‬‫ً‬‫ﺿﻐط اﻟدم ﻣﻠﺣوظﺔ ﺑﻘدر أﻛﺑر‪ .‬وﻋﺎدة ﻣﺎ ﯾرﺗﻔﻊ ﻣﺳﺗوى ﺿﻐط اﻟدم ﻓﻲ أﺛﻧﺎء اﻟﻌﻣل أو اﻟﻠﻌب‪ ،‬وﯾﻧﺧﻔض‬‫ﻛﺛﯾرا ﺣﯾﺎل اﻟﻧﺗﺎﺋﺞ اﻟﺗﻲ ﺗظﮭر ﻣن ﻋﻣﻠﯾﺔ ﻗﯾﺎس واﺣدة‪.‬‬‫إﻟﻰ أدﻧﻰ ﻣﺳﺗوﯾﺎﺗﮫ ﻓﻲ أﺛﻧﺎء اﻟﻧوم‪ .‬ﻟذﻟك‪ ،‬ﻻ ﺗﻘﻠﻖ‬‫ً‬‫اﻟﻌرﺑﯾﺔ ‪20‬‬‫أﺟر ﻋﻣﻠﯾﺎت اﻟﻘﯾﺎس ﻓﻲ اﻟوﻗت ﻧﻔﺳﮫ ﻛل ﯾوم ﺑﺎﺗﺑﺎع‬‫ِ‬‫اﻹﺟراء اﻟﻣوﺿّﺢ ﻓﻲ ھذا اﻟدﻟﯾل ﻟﻣﻌرﻓﺔ اﻟﻣﻌدل‬‫اﻟطﺑﯾﻌﻲ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪ .‬وﺗﻘدّم اﻟﻘراءات‬‫اﻟﻣﻧﺗظﻣﺔ ﺳﺟﻼً أﻛﺛر ﺷﻣوﻟﯾﺔ ﻟﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫اﺣرص ﻋﻠﻰ ﺗوﺿﯾﺢ اﻟوﻗت واﻟﺗﺎرﯾﺦ ﻋﻧد ﺗﺳﺟﯾل‬‫ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪ .‬اﺳﺗﺷر اﻟطﺑﯾب ﻟﯾﻔﺳّر ﻟك‬‫ﺑﯾﺎﻧﺎت ﻣﺳﺗوى ﺿﻐط اﻟدم‪.‬‬‫اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ‬‫اﻟﻣﺷﻛﻠﺔ‬‫ﻻ ﯾظﮭر ﺷﻲء ﻋﻠﻰ‬‫اﻟﺷﺎﺷﺔ‪ ،‬ﺣﺗﻰ ﻋﻧد‬‫ﺗﺷﻐﯾل اﻟﺟﮭﺎز‪.‬‬‫ﻻ ﯾﻧﺗﻔﺦ اﻟرﺑﺎط‬‫ﺑﺎﻟﮭواء‪.‬‬‫اﻟﺟﮭﺎز ﻻ ﯾﻘﯾس‪.‬‬‫اﻟﻘراءات ﻣرﺗﻔﻌﺔ أو‬‫ﻣﻧﺧﻔﺿﺔ ﻟﻠﻐﺎﯾﺔ‪.‬‬‫اﻟﺳﺑب اﻟﻣﺣﺗﻣل‬‫ﻧﻔدت طﺎﻗﺔ اﻟﺑطﺎرﯾﺎت‪.‬‬‫أطراف اﻟﺑطﺎرﯾﺎت ﻟﯾﺳت ﻓﻲ‬‫ﻣواﺿﻌﮭﺎ اﻟﺻﺣﯾﺣﺔ‪.‬‬‫ﺟﮭد اﻟﺑطﺎرﯾﺔ ﻣﻧﺧﻔض ﻟﻠﻐﺎﯾﺔ‪.‬‬‫)رﻣز ‪LOW‬‬‫ﯾوﻣض‬‫‪ .(BATTERY‬إذا ﻧﻔدت‬‫اﻟﺑطﺎرﯾﺎت ﺗﻣﺎ ًﻣﺎ‪ ،‬ﻓﻠن ﯾظﮭر اﻟرﻣز‪.‬‬‫ﻟم ﯾُﻠف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﻣﻼﺋﻣﺔ‪.‬‬‫ﺣرﻛت ذراﻋك أو ﺟﺳدك ﻓﻲ أﺛﻧﺎء‬‫ّ‬‫اﻟﻘﯾﺎس‪.‬‬‫ﻣوﺿﻊ اﻟرﺑﺎط ﻏﯾر ﺻﺣﯾﺢ‪.‬‬‫اﻟﻘﯾﻣﺔ ﻣﺧﺗﻠﻔﺔ ﻋن اﻟﻘﯾﺎس ﻓﻲ اﻟﻌﯾﺎدة‬‫أو ﻟدى اﻟطﺑﯾب‪.‬‬‫اﻹﺟراء اﻟﻣوﺻﻰ ﺑﮫ‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪.‬‬‫أ ِﻋد ﺗرﻛﯾب اﻟﺑطﺎرﯾﺎت ﺑﺣﯾث ﺗﻛون اﻷطراف‬‫اﻟﺳﺎﻟﺑﺔ واﻟﻣوﺟﺑﺔ ﻣﺗطﺎﺑﻘﺔ ﻣﻊ ﺗﻠك اﻟﻣوﺿﺣﺔ‬‫ﻓﻲ ﺣﺟرة اﻟﺑطﺎرﯾﺎت‪.‬‬‫اﺳﺗﺑدل ﺟﻣﯾﻊ اﻟﺑطﺎرﯾﺎت ﺑﺄﺧرى ﺟدﯾدة‪.‬‬‫ﻟف اﻟرﺑﺎط ﺑطرﯾﻘﺔ ﺻﺣﯾﺣﺔ‪.‬‬‫اﺣرص ﻋﻠﻰ أن ﺗظل ﺛﺎﺑﺗًﺎ وھﺎدﺋًﺎ ﻓﻲ أﺛﻧﺎء‬‫اﻟﻘﯾﺎس‪.‬‬‫اﺟﻠس ﻓﻲ ﺣﺎﻟﺔ ﺛﺑﺎت وھدوء‪.‬‬‫ﺿﻊ ذراﻋك ﻋﻠﻰ طﺎوﻟﺔ ﺑﺣﯾث ﺗﻛون راﺣﺔ‬‫ﯾدك ﻣواﺟﮭﺔ ﻷﻋﻠﻰ واﻟرﺑﺎط ﻋﻧد ﻣﺳﺗوى‬‫اﻟﻘﻠب‪.‬‬‫إذا ﻛﺎن ﻣﻌدل ﺿرﺑﺎت اﻟﻘﻠب ﻟدﯾك ﺿﻌﯾﻔًﺎ ﺟدًا‬‫أو ﻏﯾر ﻣﻧﺗظم‪ ،‬ﻓﻘد ﯾﺟد اﻟﺟﮭﺎز ﺻﻌوﺑﺔ ﻓﻲ‬‫ﺗﺣدﯾد ﻣﺳﺗوى ﺿﻐط اﻟدم ﻟدﯾك‪.‬‬‫راﺟﻊ "ﻣﺎ ﻓﺎﺋدة ﻗﯾﺎس ﺿﻐط اﻟدم ﻓﻲ اﻟﻣﻧزل؟"‪.‬‬‫أﺧرى‬‫اﻧزع اﻟﺑطﺎرﯾﺎت‪ .‬ﺿﻌﮭﺎ ﻣرة أﺧرى ﺑﺷﻛل‬‫ﺻﺣﯾﺢ و أﻋد إﺟراء اﻟﻘﯾﺎس ﻣرة أﺧرى‪.‬‬‫ﻣﻼﺣظﺔ‪ :‬إذا ﻟم ﺗُﺣل اﻟﻣﺷﻛﻼت ﺑﺎﺗﺑﺎع اﻹﺟراءات اﻟﻣوﺿﺣﺔ أﻋﻼه‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل‪ .‬ﻻ ﺗﺣﺎول ﻓﺗﺢ أو إﺻﻼح‬‫ﻧظرا إﻟﻰ أن أي ﻣﺣﺎوﻟﺔ ﻟﻠﻘﯾﺎم ﺑذﻟك ﺳﺗؤدي إﻟﻰ إﻟﻐﺎء اﻟﺿﻣﺎن‪.‬‬‫ھذا اﻟﻣﻧﺗﺢ‪ً ،‬‬‫اﻟﻌرﺑﯾﺔ ‪21‬‬‫اﻟﺻﯾﺎﻧﺔ‬‫ﻣﻛوﻧﺎت ﻛﮭرﺑﺎﺋﯾﺔ ﺷدﯾدة اﻟﺣﺳﺎﺳﯾﺔ ووﺣدة ھواء ﻣﻌﻘدة ﯾﻣﻛن أن ﺗﺗﻠف‪ .‬وإذا ﺗﻌذّر‬‫ﻻ ﺗﻔﺗﺢ اﻟﺟﮭﺎز‪ .‬ﺗُﺳﺗﺧدم ّ‬‫ﻋﻠﯾك ﺣل اﻟﻣﺷﻛﻠﺔ ﺑﺎﺗﺑﺎع إرﺷﺎدات اﺳﺗﻛﺷﺎف اﻷﺧطﺎء وإﺻﻼﺣﮭﺎ‪ ،‬ﻓﺎﺗﺻل ﺑﺎﻟوﻛﯾل اﻟﻣﻌﺗﻣد ﻓﻲ ﻣﻧطﻘﺗك أو‬‫ﺳﯾزود ﻗﺳم ﺧدﻣﺔ اﻟﻌﻣﻼء اﻟﺗﺎﺑﻊ ﻟﺷرﻛﺔ ‪ A&D‬اﻟﻣﻌﻠوﻣﺎت اﻟﻔﻧﯾﺔ وﻗطﻊ اﻟﻐﯾﺎر‬‫ﻗﺳم ﺧدﻣﺔ اﻟﻌﻣﻼء‪.‬‬‫ّ‬‫واﻟوﺣدات ﻟﻠوﻛﻼء اﻟﻣﻌﺗﻣدﯾن‪.‬‬‫ﺻﻧّﻊ ﻟﯾﺗﻣﺗﻊ ﺑﻔﺗرة ﺗﺷﻐﯾل طوﯾﻠﺔ‪ .‬وﻣﻊ ذﻟك‪ ،‬ﻓﺈﻧﮫ ﯾوﺻﻰ ﺑﺷﻛل ﻋﺎم ﺑﻔﺣص اﻟﺟﮭﺎز ﻛل‬‫ﺻﻣم اﻟﺟﮭﺎز و ُ‬‫ُ‬‫ﻋﺎﻣﯾن ﻟﺿﻣﺎن اﻟﺗﺷﻐﯾل اﻟﺻﺣﯾﺢ ﻟﮫ ودﻗﺔ أداﺋﮫ‪ .‬ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑﺎﻟوﻛﯾل اﻟﻣﻌﺗﻣد ﻓﻲ ﻣﻧطﻘﺗك أو ﺷرﻛﺔ‬‫‪ A&D‬ﻹﺟراء اﻟﺻﯾﺎﻧﺔ‪.‬‬‫اﻟﺑﯾﺎﻧﺎت اﻟﻔﻧﯾﺔ‬‫اﻟﻧوع‬‫‪UA-1020-W‬‬‫طرﯾﻘﺔ اﻟﻘﯾﺎس‬‫اﻟﻘﯾﺎس ﺑﺎﻟذﺑذﺑﺎت‬‫ﻧطﺎق اﻟﻘﯾﺎس‬‫‪mmHg 299 - 0‬‬‫اﻟﺿﻐط‪:‬‬‫اﻟﺿﻐط اﻻﻧﻘﺑﺎﺿﻲ‪mmHg 279 - 60 :‬‬‫اﻟﺿﻐط اﻻﻧﺑﺳﺎطﻲ‪mmHg 200 - 40 :‬‬‫‪ 180 - 40‬ﻧﺑﺿﺔ‪ /‬اﻟدﻗﯾﻘﺔ‬‫اﻟﻧﺑض‪:‬‬‫دﻗﺔ اﻟﻘﯾﺎس‬‫اﻟﺿﻐط‪:‬‬‫اﻟﻧﺑض‪:‬‬‫‪mmHg 3±‬‬‫‪%5±‬‬‫ﻣﺻدر اﻹﻣداد ﺑﺎﻟطﺎﻗﺔ‬‫‪ 4‬ﺑطﺎرﯾﺎت ﺑﺟﮭد ‪ 1.5‬ﻓوﻟت )ﻣﻘﺎس ‪ ،R6P‬أو ‪ ،LR6‬أو ‪،AA) H‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد )‪) (TB-233C‬ﻏﯾر ﻣرﻓﻖ(‬‫ّ‬‫ﺗﻘرﯾﺑًﺎ ‪ 1,000‬ﻣرة ‪) LR6‬ﺑطﺎرﯾﺎت ﻗﻠوﯾﺔ(‬‫اﻟﺑطﺎرﯾﺔ ﻣﻘﺎس ‪) R6P‬ﺑطﺎرﯾﺎت ﻣﻧﺟﻧﯾز(‬‫‪ 300‬ﻣرة ﻗﯾﺎس ﺗﻘرﯾﺑًﺎ‬‫ﻣﻊ ﻗﯾﻣﺔ ﺿﻐط ﺗﺳﺎوي ‪ mmHg 180‬ﻓﻲ ﻏرﻓﺔ درﺟﺔ ﺣرارﺗﮭﺎ‬‫‪ °23‬ﻣﺋوﯾﺔ‪.‬‬‫اﻟﺗﺻﻧﯾف‬‫ﻣﻌدات ‪ ME‬ﻣزودة ﺑﺎﻟطﺎﻗﺔ داﺧﻠﯾًﺎ )ﻣزودة ﺑﺎﻟﺑطﺎرﯾﺎت(‬‫ﺑﻣﺣول ﺗﯾﺎر(‬‫اﻟﺗﺻﻧﯾف ‪) II‬ﻣزودة‬‫ّ‬‫وﺿﻊ اﻟﺗﺷﻐﯾل اﻟﻣﺳﺗﻣر‬‫وﻓﻘًﺎ ﻟﺷﮭﺎدة ‪ISO81060-2：2013‬‬‫ﻓﻲ دراﺳﺔ اﻟﺗﺣﻘﻖ ﻣن ﺻﺣﺔ اﻟﺗﺟﺎرب اﻟﺳرﯾرﯾﺔ‪ ،‬ﻛﺎن ‪ K5‬ﯾﺳﺗﺧدم ﻓﻲ‬‫ﻋﺎ ﻟﺗﺣدﯾد ﺿﻐط اﻟدم اﻻﻧﺑﺳﺎطﻲ‪.‬‬‫‪ 85‬ﻣوﺿو ً‬‫ﻋدد ﻣرات اﻟﻘﯾﺎس‬‫اﻻﺧﺗﺑﺎر اﻟﺳرﯾري‬‫‪EMD‬‬‫‪IEC 60601-1-2: 2014‬‬‫اﻟذاﻛرة‬‫آﺧر ‪ 90‬ﻋﻣﻠﯾﺔ ﻗﯾﺎس‬‫ظروف اﻟﺗﺷﻐﯾل‬‫ظروف اﻟﻧﻘل ‪ /‬اﻟﺗﺧزﯾن‬‫اﻷﺑﻌﺎد‬‫‪ +10‬إﻟﻰ ‪ 15 / C° +40‬إﻟﻰ ‪ 800 / RH% 85‬إﻟﻰ ‪hPa 1,060‬‬‫‪ -20‬إﻟﻰ ‪ 10 / C° +60‬إﻟﻰ ‪ 700 / RH% 95‬إﻟﻰ ‪hPa 1,060‬‬‫ﺗﻘرﯾﺑًﺎ ‪] 140‬ﻋرض[ × ‪] 60‬ارﺗﻔﺎع[ × ‪] 105‬ﻋﻣﻖ[ ﻣم‬‫اﻟوزن‬‫ﺗﻘرﯾﺑًﺎ ‪ 285‬ﺟرا ًﻣﺎ دون اﻟﺑطﺎرﯾﺎت‬‫اﻟﻌرﺑﯾﺔ ‪22‬‬‫ﻧوع اﻟرﺑﺎط ‪BF‬‬‫اﻟﺟزء اﻟﻣﺳﺗﺧدم‬‫ﺟﮭﺎز ﺣﻣﺎﯾﺔ اﻟﻣدﺧل‪IP20 :‬‬‫ﺟﮭﺎز ﺣﯾﺎﺗﻲ‬‫ﻣﻔﯾد‪ 5 :‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫اﻟرﺑﺎط‪ 2 :‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‪ 5 :‬ﺳﻧوات )ﺑﺎﺳﺗﺧداﻣﮫ ﺳت ﻣرات ﻓﻲ اﻟﯾوم(‬‫ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ُﻣﻠﺣﻖ ّ‬‫اﻟﻣﺣول ﻟﺗوﺻﯾل ﺟﮭﺎز ﻗﯾﺎس ﺿﻐط اﻟدم ﺑﻣﺻدر طﺎﻗﺔ ﻓﻲ اﻟﻣﻧزل‪.‬‬‫ّ‬‫‪TB-233C‬‬‫ﯾُرﺟﻰ اﻻﺗﺻﺎل ﺑوﻛﯾل ‪ A&D‬اﻟﻣﺣﻠﻲ ﻟﺷراﺋﮫ‪.‬‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد أو اﺳﺗﺑداﻟﮫ ﺑﺷﻛل دوري‪.‬‬‫ﯾﻛون ﺿرورﯾًﺎ ﻓﺣص ّ‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟرﻣوز اﻟﻣطﺑوﻋﺔ ﻋﻠﻰ ّ‬‫اﻟوظﯾﻔﺔ ‪ /‬اﻟﻣﻌﻧﻰ‬‫اﻟرﻣوز‬‫ﻟﻼﺳﺗﺧدام اﻟﻣﻧزﻟﻲ ﻓﻘط‬‫اﻟﺟﮭﺎز ﻣن اﻟﻔﺋﺔ ‪II‬‬‫اﻟﻣﺻﮭر اﻟﺣراري‬‫اﻟﻣﺻﮭر‬‫ﻣﻠﺻﻖ اﻟﺗوﺟﯾﮫ ‪ EC‬ﻟﻸﺟﮭزة‬‫ﻣﻠﺻﻖ ﺷﮭﺎدة ‪ EAC‬ﻟﻸﺟﮭزة‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫اﻟﻘطﺑﯾﺔ ﻟﻘﺎﺑس ّ‬‫اﻟﻣﻠﺣﻘﺎت ﺗُﺑﺎع ﺑﺷﻛل ﻣﻧﻔﺻل‬‫اﻟرﺑﺎط‬‫ﻣﺣول اﻟﺗﯾﺎر اﻟﻣﺗردد‬‫ّ‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫ﺣﺟم اﻟرﺑﺎط‬‫ﺣﺟم اﻟذراع‬‫‪CUF-F-LA‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﻛﺑﯾر‬‫‪ 31‬ﺳم إﻟﻰ ‪ 45‬ﺳم‬‫‪CUF-I‬‬‫اﻟرﺑﺎط ذو اﻟﻧطﺎق‬‫اﻟﻌرﯾض‬‫‪ 22‬ﺳم إﻟﻰ ‪ 42‬ﺳم‬‫‪CUF-F-A‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن‬‫‪ 22‬ﺳم إﻟﻰ ‪ 32‬ﺳم‬‫‪CUF-F-SA‬‬‫رﺑﺎط اﻟﺑﺎﻟﻐﯾن اﻟﺻﻐﯾر‬‫‪ 16‬ﺳم إﻟﻰ ‪ 24‬ﺳم‬‫رﻗم اﻟﻛﺗﺎﻟوج‬‫‪TB-233C‬‬‫ﻗﺎﺑس‬‫اﻟﻧوع ‪C‬‬

‫ﻣﻼﺣظﺔ‪ :‬ﺗﺧﺿﻊ اﻟﻣواﺻﻔﺎت ﻟﻠﺗﻐﯾﯾر دون إﺷﻌﺎر ﻣﺳﺑﻖ‪.‬‬‫ﺗﺻﻧﯾف ‪ IP‬ھو درﺟﺎت اﻟﺣﻣﺎﯾﺔ اﻟﺗﻲ ﯾﺗﻣﺗﻊ ﺑﮭﺎ اﻟﺟﮭﺎز وﻓﻘًﺎ ﻟﻠﺗوﺟﯾﮫ ‪ .IEC 60529‬ھذا اﻟﺟﮭﺎز‬‫ﻣﺣﻣﻲ ﺿد اﻟﻣواد اﻟﻐرﯾﺑﺔ اﻟﺻﻠﺑﺔ اﻟﺗﻲ ﯾﺑﻠﻎ ﻗطرھﺎ ‪ 12‬ﻣم أو أﻛﺑر‪ ،‬ﻣﺛل اﻹﺻﺑﻊ‪ .‬ھذا اﻟﺟﮭﺎز ﻏﯾر‬‫ﻣﺣﻣﻲ ﺿد اﻟﻣﺎء‪.‬‬‫اﻟﻌرﺑﯾﺔ ‪23‬‬‫ﻣذﻛرة‬‫اﻟﻌرﺑﯾﺔ ‪24‬‬‫ﻓﮭرﺳت ﻣﻧدرﺟﺎت‬‫ﻣﺷﺗرﯾﺎن ﮔراﻣﯽ‪2 ....................... ................................................................‬‬‫ﺗذﮐرات اوﻟﯾﮫ‪2 .......................... ................................................................‬‬‫اﻗداﻣﺎت اﺣﺗﯾﺎطﯽ‪2 ...................... ................................................................‬‬‫ﻣﻌرﻓﯽ ﻗطﻌﺎت ‪4 ........................ ................................................................‬‬‫ﻧﻣﺎدھﺎ‪5.. ................................................................................................‬‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت ‪7 ......................... ................................................................‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‪8 ............... ................................................................‬‬‫ﻧﺼﺐ ‪ /‬ﺗﻌﻮﯾﺾ ﺑﺎﺗﺮیھﺎ ‪8 ....... ................................................................‬‬‫ﻣﺘﺼﻞ ﮐﺮدن ﺷﯿﻠﻨﮓ ھﻮا‪9 ........ ................................................................‬‬‫ﻣﺘﺼﻞ ﮐﺮدن آداﭘﺘﻮر ‪9 ...... ................................................................ AC‬‬‫ﺗﻨﻈﯿﻢ ﺳﺎﻋﺖ داﺧﻠﯽ‪10 ............ ................................................................‬‬‫اﻧﺘﺨﺎب ﺳﺎﯾﺰ ﻣﻨﺎﺳﺐ ﺑﺎزوﺑﻨﺪ‪11 .. ................................................................‬‬‫اﺳﺘﻔﺎده از ﺑﺎزوﺑﻨﺪ ‪11 ............. ................................................................‬‬‫ﻧﺤﻮه اﻧﺠﺎم اﻧﺪازهﮔﯿﺮی دﻗﯿﻖ‪13 .. ................................................................‬‬‫اﻧﺪازهﮔﯿﺮی ‪13 .................... ................................................................‬‬‫ﭘﺲ از اﻧﺪازهﮔﯿﺮی ‪13 ............ ................................................................‬‬‫اﻧدازهﮔﯾریھﺎ ‪14 ....................... ................................................................‬‬‫اﻧﺪازهﮔﯿﺮی ﻋﺎدی ‪14 ............. ................................................................‬‬‫اﻧﺪازهﮔﯿﺮی ﺑﺎ ﻓﺸﺎر ﺗﻨﻈﯿﻢ ﺷﺪه‪15 ................................................................‬‬‫اﻧﺪازهﮔﯿﺮی ﺑﺎ ﻓﺸﺎر ﺳﯿﺴﺘﻮﻟﯽ ﻣﻄﻠﻮب‪15 ........................ ................................‬‬‫ﻧﮑﺎﺗﯽ ﺑﺮای اﻧﺪازهﮔﯿﺮی دﻗﯿﻖ‪15. ................................................................‬‬‫اﻧدازهﮔﯾری ‪16 ....... ................................................................ TriCheckTM‬‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ‪18 ......... ................................................................‬‬‫ﺷﺎﺧص ‪ IHB/AFib‬ﭼﯾﺳت؟ ‪19 ..... ................................................................‬‬‫‪ AFib‬ﭼﯾﺳت؟‪19 ...................... ................................................................‬‬‫ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر ‪19 .................. ................................................................‬‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪19 ............. ................................(WHO‬‬‫در ﻣورد ﻓﺷﺎر ﺧون‪20 ................ ................................................................‬‬‫ﻓﺸﺎر ﺧﻮن ﭼﯿﺴﺖ؟ ‪20 ............ ................................................................‬‬‫ﻓﺸﺎر ﺧﻮن ﺑﺎﻻ ﭼﯿﺴﺖ و ﭼﮕﻮﻧﮫ ﮐﻨﺘﺮل ﻣﯽﺷﻮد؟ ‪20 ............ ................................‬‬‫ﭼﺮا ﻓﺸﺎر ﺧﻮن را در ﺧﺎﻧﮫ اﻧﺪازهﮔﯿﺮی ﮐﻨﻢ؟ ‪20 ................ ................................‬‬‫طﺒﻘﮫﺑﻨﺪی ﻓﺸﺎر ﺧﻮن ‪20. ................................................................WHO‬‬‫ﻧﻮﺳﺎﻧﺎت ﻓﺸﺎر ﺧﻮن ‪20 ........... ................................................................‬‬‫ﻋﯾبﯾﺎﺑﯽ‪21 .............................................................................................‬‬‫ﻧﮕﮭداری ‪22 ............................................................................................‬‬‫دادهھﺎی ﻓﻧﯽ‪22 ......................... ................................................................‬‬‫ﻓﺎرﺳﯽ ‪1‬‬‫ﻣﺷﺗرﯾﺎن ﮔراﻣﯽ‬‫ﺑﺎﺑت ﺧرﯾد ﺟدﯾدﺗرﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ‪ A&D‬ﮐﮫ ﯾﮑﯽ از ﭘﯾﺷرﻓﺗﮫﺗرﯾن دﺳﺗﮕﺎهھﺎی ﭘﺎﯾش ﻣوﺟود‬‫در ﺑﺎزار اﺳت‪ ،‬ﺑﮫ ﺷﻣﺎ ﺗﺑرﯾﮏ ﻣﯽﮔوﯾﯾم‪ .‬اﯾن دﺳﺗﮕﺎه ﮐﮫ ﺑرای اﺳﺗﻔﺎده آﺳﺎن و در ﻋﯾن ﺣﺎل دﻗﯾﻖ طراﺣﯽ‬‫ﺷده اﺳت‪ ،‬ﺑررﺳﯽ روزاﻧﮫ ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﺳﮭﯾل ﻣﯽﮐﻧد‪.‬‬‫ﺗوﺻﯾﮫ ﻣﯽﺷود ﭘﯾش از آﻧﮑﮫ ﺑرای اوﻟﯾن ﺑﺎر از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬اﯾن دﻓﺗرﭼﮫ راھﻧﻣﺎ را‬‫ﺑﮫدﻗت ﻣطﺎﻟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﺗذﮐرات اوﻟﯾﮫ‬‫اﯾن دﺳﺗﮕﺎه ﺑﺎ دﺳﺗوراﻟﻌﻣل ‪ 93/42 EEC‬ﺑرای ﻣﺣﺻوﻻت ﭘزﺷﮑﯽ در اروﭘﺎ ﻣطﺎﺑﻘت دارد‪ .‬ﻧﺷﺎن‬‫اﻧطﺑﺎق‬‫ﺷﺎھدی ﺑر اﯾن ﻣدﻋﯽ اﺳت‪.‬‬‫)‪ :0123‬ﺷﻣﺎره ارﺟﺎع ﺑﮫ ﺑﺧش اﻋﻼم ﺷده ﻣرﺑوطﮫ(‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﺳﺗﻔﺎده ﺑزرﮔﺳﺎﻻن طراﺣﯽ ﺷده اﺳت‪ ،‬ﻧﮫ ﻧوزادان ﯾﺎ اطﻔﺎل‪.‬‬‫ﻣﺣﯾط ﮐﺎرﺑرد‪ .‬اﯾن دﺳﺗﮕﺎه ﺑﮫﮔوﻧﮫای طراﺣﯽ ﺷده اﺳت ﮐﮫ ﺷﻣﺎ ﺑﺗواﻧﯾد در ﻣﺣﯾط ﻣﻧزﻟﺗﺎن ﺷﺧﺻﺎ ً از آن‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون و ﺿرﺑﺎن ﻗﻠب اﻓراد ﺑرای ﺗﺷﺧﯾص ﺑﯾﻣﺎری طراﺣﯽ ﺷده اﺳت‪.‬‬‫اﻗداﻣﺎت اﺣﺗﯾﺎطﯽ‬‫در ﺳﺎﺧت اﯾن دﺳﺗﮕﺎه از اﺟزای دﻗﯾﻖ اﺳﺗﻔﺎده ﺷده اﺳت‪ .‬دﻣﺎ و رطوﺑت ﺷدﯾد‪ ،‬ﻧور ﻣﺳﺗﻘﯾم ﺧورﺷﯾد‪،‬‬‫ﺿرﺑﮫ ﯾﺎ ﮔرد و ﺧﺎک ﻣﯽﺗواﻧد ﺑﮫ آن آﺳﯾب ﺑرﺳﺎﻧد‪.‬‬‫ﺑﺎزوﺑﻧد و دﺳﺗﮕﺎه را ﺑﺎ ﯾﮏ ﭘﺎرﭼﮫ ﻧرم و ﺧﺷﮏ ﯾﺎ ﭘﺎرﭼﮫ ﻧﻣدار ﺷده ﺑﺎ آب ﯾﺎ ﺷوﯾﻧده ﺧﻧﺛﯽ ﺗﻣﯾز ﮐﻧﯾد‪.‬‬‫ھرﮔز از اﻟﮑل‪ ،‬ﺑﻧزن‪ ،‬ﺗﯾﻧر ﯾﺎ دﯾﮕر ﻣواد ﺷﯾﻣﯾﺎﯾﯽ ﺧورﻧده ﺑرای ﺗﻣﯾز ﮐردن ﺑﺎزوﺑﻧد ﯾﺎ دﺳﺗﮕﺎه اﺳﺗﻔﺎده‬‫ﻧﮑﻧﯾد‪.‬‬‫از ﺧم ﮐردن زﯾﺎد ﺑﺎزوﺑﻧد ﯾﺎ ﭘﯾﭼﺎﻧدن ﺷﯾﻠﻧﮓ ﺑرای ﻣدت طوﻻﻧﯽ اﺟﺗﻧﺎب ﮐﻧﯾد‪ ،‬زﯾرا اﯾن ﮐﺎر ﻣوﺟب ﮐوﺗﺎه‬‫ﺷدن ﻋﻣر اﺟزای دﺳﺗﮕﺎه ﻣﯽﺷود‪.‬‬‫ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ ﺷﯾﻠﻧﮓ و ﮐﺎﺑل دﺳﺗﮕﺎه ﺑﮫ دور ﮔردن ﮐودﮐﺎن ﯾﺎ ﻧوزادان ﻧﭘﯾﭼد‪ ،‬زﯾرا ﻣﯽﺗواﻧد ﻣوﺟب‬‫ﺧﻔﮕﯽ آﻧﮭﺎ ﺷود‪.‬‬‫ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ در ﺣﯾن اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺷﯾﻠﻧﮓ ھوا دﭼﺎر ﭘﯾﭻﺧوردﮔﯽ ﻧﺷود‪ .‬اﯾن اﻣر ﻣﯽﺗواﻧد ﺑر‬‫اﺛر ﻓﺷﺎر ﭘﯾوﺳﺗﮫ ﺑﺎزوﺑﻧد ﺑﺎﻋث اﯾﺟﺎد آﺳﯾب ﺷود‪.‬‬‫دﺳﺗﮕﺎه و ﺑﺎزوﺑﻧد در ﺑراﺑر آب ﻣﻘﺎوم ﻧﯾﺳﺗﻧد‪ .‬ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ آب‪ ،‬ﺑﺎران و ﻋرق روی دﺳﺗﮕﺎه و ﺑﺎزوﺑﻧد‬‫ﻟﮑﮫ اﯾﺟﺎد ﻧﮑﻧﻧد‪.‬‬‫اﮔر از دﺳﺗﮕﺎه در ﻧزدﯾﮑﯽ ﺗﻠوﯾزﯾون‪ ،‬ﻓر ﻣﺎﯾﮑرووﯾو‪ ،‬ﺗﻠﻔن ھﻣراه‪ ،‬اﺷﻌﮫ اﯾﮑس ﯾﺎ در ﻧزدﯾﮑﯽ دﯾﮕر‬‫اﺑزارھﺎﯾﯽ اﺳﺗﻔﺎده ﮐﻧﯾد ﮐﮫ ﻣﯾدان اﻟﮑﺗرﯾﮑﯽ ﻗوی دارﻧد‪ ،‬ﻣﻣﮑن اﺳت در اﻧدازهﮔﯾری اﺧﺗﻼل اﯾﺟﺎد ﺷود‪.‬‬‫دﺳﺗﮕﺎهھﺎی ارﺗﺑﺎطﯽ ﺑﯽﺳﯾم ﻣﺎﻧﻧد دﺳﺗﮕﺎهھﺎی ﺷﺑﮑﮫ ﺧﺎﻧﮕﯽ و ﺗﻠﻔنھﺎی ھﻣراه‪ ،‬ﺗﻠﻔنھﺎی ﺑﯽﺳﯾم و ﭘﺎﯾﮫ‬‫آﻧﮭﺎ‪ ،‬و ﺑﯾﺳﯾمھﺎ ﻣﯽﺗواﻧﻧد ﺑر اﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺗﺄﺛﯾر ﺑﮕذارﻧد‪.‬‬‫ﻟذا ﻻزم اﺳت ﮐﮫ ﺣداﻗل ‪ 30‬ﺳﺎﻧﺗﯽ ﻣﺗر از اﯾن دﺳﺗﮕﺎهھﺎ ﻓﺎﺻﻠﮫ داﺷﺗﮫ ﺑﺎﺷﯾد‪.‬‬‫ﺗﺟﮭﯾزات‪ ،‬ﻗطﻌﺎت و ﺑﺎﺗریھﺎﯾﯽ ﮐﮫ اﺳﺗﻔﺎده ﺷدهاﻧد‪ ،‬ﻧﺑﺎﯾد ﺑﮫﻋﻧوان ﺿﺎﯾﻌﺎت ﻋﺎدی ﺧﺎﻧﮕﯽ ﻗﻠﻣداد ﺷوﻧد‪ ،‬و‬‫ﺑﺎﯾد ﺑر اﺳﺎس ﻣﻘررات ﻣﺣﻠﯽ دور اﻧداﺧﺗﮫ ﺷوﻧد‪.‬‬‫وﻗﺗﯽ از آداﭘﺗور ‪ AC‬اﺳﺗﻔﺎده ﻣﯽﮐﻧﯾد‪ ،‬ﻣطﻣﺋن ﺷوﯾد ﮐﮫ آداﭘﺗور ‪ AC‬ﻣﯽﺗواﻧد در ﺻورت ﻟزوم ﺑﮫ‬‫ﺳﮭوﻟت از ﭘرﯾز ﺑرق ﺧﺎرج ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪2‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد ﻣﺟددا ً از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬از ﺗﻣﯾز ﺑودن دﺳﺗﮕﺎه ﻣطﻣﺋن ﺷوﯾد‪.‬‬‫ﺗﻐﯾﯾری در دﺳﺗﮕﺎه اﯾﺟﺎد ﻧﮑﻧﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺑﮫ دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﺑﮫﻣﻧظور اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪ ،‬ﺑﺎزو ﺑﺎﯾد ﺗوﺳط ﺑﺎزوﺑﻧد ﺑﮫﻗدری ﻓﺷرده ﺷود ﮐﮫ ﻣوﻗﺗﺎ ً ﺟرﯾﺎن ﺧون از‬‫ﻣﯾﺎن ﺳرﺧرگ را ﻣﺗوﻗف ﮐﻧد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد درد‪ ،‬ﮐرﺧﺗﯽ ﯾﺎ ﻗرﻣزی ﻣوﻗت ﺑﺎزو ﺷود‪.‬‬‫ﭼﻧﯾن ﺷراﯾطﯽ ﺑﮫوﯾژه زﻣﺎﻧﯽ اﯾﺟﺎد ﻣﯽﺷود ﮐﮫ اﻧدازهﮔﯾری را ﭼﻧد ﻣرﺗﺑﮫ ﭘﺷت ﺳرھم اﻧﺟﺎم ﻣﯽدھﯾد‪.‬‬‫ھرﮔوﻧﮫ درد‪ ،‬ﮐرﺧﺗﯽ ﯾﺎ ﻗرﻣزی ﺑﺎ ﮔذﺷت زﻣﺎن از ﺑﯾن ﻣﯽرود‪.‬‬‫اﮔر اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﭼﻧدﯾن ﻣرﺗﺑﮫ ﺗﮑرار ﺷود‪ ،‬ﻣﯽﺗواﻧد ﺑﮫ دﻟﯾل ﺗداﺧﻠﯽ ﮐﮫ در ﺟرﯾﺎن ﺧون اﯾﺟﺎد‬‫ﻣﯽﮐﻧد‪ ،‬ﺑﮫ ﻓرد آﺳﯾب ﺑزﻧد‪ .‬اﯾن ﻣوﺿوع را ﺑررﺳﯽ ﮐﻧﯾد ﮐﮫ در ﺻورت ﺗﮑرار اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪،‬‬‫ﻋﻣﻠﯾﺎت دﺳﺗﮕﺎه ﻣوﺟب اﺧﺗﻼل طوﻻﻧﯽ ﻣدت در ﺟرﯾﺎن ﺧون ﻧﺷود‪.‬‬‫اﮔر ﻋﻣل ﻣﺎﺳﺗﮑﺗوﻣﯽ داﺷﺗﮫاﯾد و ﺳﯾﻧﮫﺗﺎن ﺗﺧﻠﯾﮫ ﺷده اﺳت‪ ،‬ﭘﯾش از اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺑﺎ ﭘزﺷﮑﺗﺎن‬‫ﻣﺷورت ﮐﻧﯾد‪.‬‬‫اﺟﺎزه ﻧدھﯾد ﮐﮫ ﮐودﮐﺎن از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﻧد و در ﺟﺎﯾﯽ ﮐﮫ در دﺳﺗرس اطﻔﺎل اﺳت از اﯾن دﺳﺗﮕﺎه‬‫اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪ .‬ﻋدم ﺗوﺟﮫ ﺑﮫ اﯾن ﻣوﺿوع ﻣﯽﺗواﻧد ﺑﺎﻋث ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺷود‪.‬‬‫اﺟزای ﮐوﭼﮑﯽ در اﯾن دﺳﺗﮕﺎه وﺟود دارد ﮐﮫ در ﺻورت ﺑﻠﻌﯾده ﺷدن ﺗوﺳط اطﻔﺎل ﻣﯽﺗواﻧد ﺑﺎﻋث ﺧﻔﮕﯽ‬‫آﻧﮭﺎ ﺷود‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ دﯾﮕر ﻧﻣﯽﺧواھﯾد از دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬آداﭘﺗور ‪ AC‬را ﺧﺎرج ﮐﻧﯾد‪.‬‬‫اﺳﺗﻔﺎده از ﻟوازم ﺟﺎﻧﺑﯽ ﮐﮫ در اﯾن راھﻧﻣﺎ ﺑﮫ آﻧﮭﺎ اﺷﺎره ﻧﺷده اﺳت ﻣﯽﺗواﻧد ﺑﺎﻋث آﺳﯾب ﺑﮫ ﻓرد و دﺳﺗﮕﺎه ﺷود‪.‬‬‫در ﺻورﺗﯽ ﮐﮫ ﺑﺎﺗری اﺗﺻﺎل ﮐوﺗﺎه ﺷود‪ ،‬ﻣﻣﮑن اﺳت داغ ﺷده و ﺑﺎﻋث اﯾﺟﺎد ﺳوﺧﺗﮕﯽ ﺷود‪.‬‬‫ﭘﯾش از اﺳﺗﻔﺎده )ﺣدود ﯾﮏ ﺳﺎﻋت(‪ ،‬اﺟﺎزه دھﯾد ﮐﮫ دﺳﺗﮕﺎه ﺧودش را ﺑﺎ ﻣﺣﯾط اطراف ﺳﺎزﮔﺎر ﮐﻧد‪.‬‬‫آزﻣﺎﯾش ﺑﺎﻟﯾﻧﯽ روی ﻧوزادان و زﻧﺎن ﺑﺎردار اﻧﺟﺎم ﻧﺷده اﺳت‪ .‬ﻟذا از اﯾن دﺳﺗﮕﺎه ﺑرای ﻧوزادان و زﻧﺎن‬‫ﺑﺎردار اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ‪ ،‬ﻓﯾش ‪ DC‬و ﺧود ﺑﯾﻣﺎر را ھﻣزﻣﺎن ﻟﻣس ﻧﮑﻧﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺑرقﮔرﻓﺗﮕﯽ ﺷود‪.‬‬‫از ﺑﺎد ﮐردن دﺳﺗﮕﺎه ﺑدون ﭘﯾﭼﺎﻧدن آن ﺑﮫ دور ﺑﺎزو اﺟﺗﻧﺎب ﮐﻧﯾد‪.‬‬‫ﻣوارد ﻣﻧﻊ اﺳﺗﻔﺎده‬‫در اﯾن ﺑﺧش ﻣوارد اﺣﺗﯾﺎطﯽ ﺑرای اﺳﺗﻔﺎده ﺻﺣﯾﺢ از دﺳﺗﮕﺎه ذﮐر ﺷده اﺳت‪.‬‬‫ﺑﺎزوﺑﻧد را روی ﺑﺎزوﯾﯽ ﮐﮫ ﺗﺟﮭﯾزات اﻟﮑﺗرﯾﮑﯽ ﭘزﺷﮑﯽ دﯾﮕری روی آن ﻣﺗﺻل اﺳت‪ ،‬ﺑﮫ ﮐﺎر ﻧﺑرﯾد‪ .‬در‬‫ﭼﻧﯾن ﺷراﯾطﯽ ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫﺧوﺑﯽ ﮐﺎر ﻧﮑﻧد‪.‬‬‫اﻓرادی ﮐﮫ دﭼﺎر ﻧﻘص ﺷدﯾد ﮔردش ﺧون در ﺑﺎزوھﺎﯾﺷﺎن ھﺳﺗﻧد ﺑﺎﯾد ﭘﯾش از اﺳﺗﻔﺎده از اﯾن دﺳﺗﮕﺎه ﺑﺎ‬‫ﭘزﺷﮑﺷﺎن ﻣﺷورت ﮐﻧﻧد ﺗﺎ از اﯾﺟﺎد ﻣﺷﮑﻼت ﭘزﺷﮑﯽ اﺟﺗﻧﺎب ﺷود‪.‬‬‫از ﺗﺷﺧﯾص ﺧودﺳراﻧﮫ ﺑر اﺳﺎس ﻧﺗﺎﯾﺞ اﻧدازهﮔﯾری ﺑﭘرھﯾزﯾد و از ھرﮔوﻧﮫ ﺧوددرﻣﺎﻧﯽ اﺟﺗﻧﺎب ﮐﻧﯾد‪.‬‬‫ھﻣﯾﺷﮫ ﺑرای ارزﯾﺎﺑﯽ ﻧﺗﯾﺟﮫ و ﺗﺟوﯾز درﻣﺎن ﻣﻧﺎﺳب ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪.‬‬‫ﺑﺎزوﺑﻧد دﺳﺗﮕﺎه ﻓﺷﺎر ﺧون را روی ﺑﺎزوﯾﯽ ﮐﮫ زﺧم درﻣﺎن ﻧﺷده دارد ﻗرار ﻧدھﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را روی ﺑﺎزوﯾﯽ ﮐﮫ ﺗﺣت ﺗزرﯾﻖ ورﯾدی ﯾﺎ اﻧﺗﻘﺎل ﺧون اﺳت‪ ،‬ﻗرار ﻧدھﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد‬‫ﺑﺎﻋث ﺣﺎدﺛﮫ ﯾﺎ آﺳﯾب ﺷود‪.‬‬‫در ﺟﺎﯾﯽ ﮐﮫ ﮔﺎزھﺎی اﺷﺗﻌﺎلﭘذﯾر ﻣﺎﻧﻧد ﮔﺎزھﺎی ﺑﯾﮭوﺷﯽ وﺟود دارد ﻧﺑﺎﯾد از اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﺷود‪ .‬اﯾن‬‫ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث اﻧﻔﺟﺎر ﺷود‪.‬‬‫در ﻣﺣﯾطھﺎﯾﯽ ﻣﺎﻧﻧد اﺗﺎﻗﮏ اﮐﺳﯾژن ﻓﺷﺎر ﺑﺎﻻ ﯾﺎ ﭼﺎدر اﮐﺳﯾژن ﮐﮫ اﮐﺳﯾژن ﺑﺳﯾﺎر ﻣﺗراﮐم دارﻧد‪ ،‬ﻧﺑﺎﯾد از‬‫اﯾن دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﺷود‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﺑﺎﻋث آﺗشﺳوزی ﯾﺎ اﻧﻔﺟﺎر ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪3‬‬‫ﻣﻌرﻓﯽ ﻗطﻌﺎت‬‫ﻓﯾش ‪DC‬‬‫ﻧﻣﺎﯾﺷﮕر‬‫دﮐﻣﮫ‬‫دﮐﻣﮫ‬‫دﮐﻣﮫ ‪SET‬‬‫ﻧری راﺑط ھوا‬‫دﮐﻣﮫ ‪START‬‬‫ﻣﺎدﮔﯽ ھوا‬‫ﺷﯾﻠﻧﮓ ھوا‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﺑﺎزوﺑﻧد‬‫ﻧﺷﺎﻧﮫ ﺷﺎﺧص‬‫درﭘوش ﺑﺎﺗری‬‫ﻣﺣﻔظﮫ ﺑﺎطری‬‫ﺑﺎﺗریھﺎی ‪ 1.5‬وﻟت‬‫)‪ LR6 ،R6P‬ﯾﺎ ‪(AA‬‬‫ﻧﻣﺎﯾﺷﮕر‬‫ﻣﯾﺎﻧﮕﯾن‬‫ﺣﺎﻓظﮫ‬‫ﺗﻧظﯾﻣﺎت ﻓﺷﺎر‬‫ﻧﻣﺎد ‪IHB/AFib‬‬‫ﻧﻣﺎد ﺧطﺎی اﻧدازه ﺑﺎزوﺑﻧد‬‫ﺳﯾﺳﺗوﻟﯽ ﻓﺷﺎر‬‫ﻧﻣﺎد ﺧطﺎی ﻧﺎﺷﯽ از ﺣرﮐت‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫ﻧﻣﺎد ‪TriCheckTM‬‬‫ﻋﻼﻣت ﻗﻠب‬‫ﺿرﺑﺎن ﻧﺑض‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ‬‫)‪ (WHO‬و ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر‬‫ﺷﺎﺧص ﺑﺎﺗری‬‫ﻧﻣﺎﯾش ﺳﺎﻋت‬‫ﻋﻼﻣت ‪AM / PM‬‬‫ﻓﺎرﺳﯽ ‪4‬‬‫ﻧﻣﺎدھﺎ‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی ﺟﻌﺑﮫ دﺳﺗﮕﺎه و آداﭘﺗور ‪ AC‬ﭼﺎپ ﺷده اﺳت‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫ﺣﺎﻟت آﻣﺎدهﺑﮫﮐﺎر و روﺷن ﮐردن دﺳﺗﮕﺎه‪.‬‬‫راھﻧﻣﺎی ﻧﺻب ﺑﺎﺗری‬‫ﺟرﯾﺎن ﻣﺳﺗﻘﯾم‬‫‪SN‬‬‫ﺷﻣﺎره ﺳ﷼‬‫ﺗﺎرﯾﺦ ﺗوﻟﯾد‬‫ﻧوع ‪ :BF‬دﺳﺗﮕﺎه‪ ،‬ﺑﺎزوﺑﻧد و ﻟوﻟﮫھﺎ ﺑﮫﮔوﻧﮫای طراﺣﯽ‬‫ﺷدهاﻧد ﮐﮫ از ﺑروز ﺑرقﮔرﻓﺗﮕﯽ ﺟﻠوﮔﯾری ﺷود‪.‬‬‫ﺑرﭼﺳب دﺳﺗوراﻟﻌﻣل ‪ EC‬دﺳﺗﮕﺎه ﭘزﺷﮑﯽ‬‫ﺑرﭼﺳب ‪WEEE‬‬‫ﺗوﻟﯾدﮐﻧﻧده‬‫ﻧﻣﺎﯾﻧده ‪EU‬‬‫ﻗطﺑﯾت ﻓﯾش ‪DC‬‬‫‪IP‬‬‫ﻧﻣﺎد ﺣﻔﺎظت ﺑﯾناﻟﻣﻠﻠﯽ‬‫در ﺟﺎی ﺧﺷﮏ ﻧﮕﮫداری ﮐﻧﯾد‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی ﻧﻣﺎﯾﺷﮕر ظﺎھر ﻣﯽﺷوﻧد‬‫ﻧﻣﺎدھﺎ‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‪.‬‬‫زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‪.‬‬‫ﺗﺎ ﺟﺎﯾﯽ ﮐﮫ ﻣﯽﺗواﻧﯾد ﺑﯽﺣرﮐت‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻧﺑض ﺷﻧﺎﺳﺎﯾﯽ ﺷد‪ ،‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﻧﻣﺎد ‪IHB/AFib‬‬‫زﻣﺎﻧﯽ ظﺎھر ﻣﯽﺷود ﮐﮫ ﺿرﺑﺎن ﻗﻠب ﻧﺎﻣﻧظم ﺷﻧﺎﺳﺎﯾﯽ‬‫ﺷود‪.‬‬‫اﯾن ﻧﻣﺎد زﻣﺎﻧﯽ ﮐﮫ ﺗﮑﺎنھﺎی ﺑﺳﯾﺎر آراﻣﯽ ﻣﺎﻧﻧد ﻟرزش‬‫ﺷﻧﺎﺳﺎﯾﯽ ﺷود ﻧﯾز ﻣﻣﮑن اﺳت روﺷن ﺷود‪.‬‬‫ﺣﺎﻟت ‪TriCheckTM‬‬‫اﯾن ﺣﺎﻟت ﺑﮫ طور ﺧودﮐﺎر و ﭘﯾﺎﭘﯽ‪ ،‬ﺳﮫ ﺑﺎر ﻋﻣﻠﯾﺎت‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم داده و ﻣﯾﺎﻧﮕﯾن آﻧﮭﺎ را ﻧﻣﺎﯾش‬‫ﻣﯽدھد‪.‬‬‫زﻣﺎﻧﯽ ظﺎھرا ً ﻣﯽﺷود ﮐﮫ ﺣرﮐﺗﯽ در ﺑدن ﯾﺎ ﺑﺎزو‬‫ﺷﻧﺎﺳﺎﯾﯽ ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪5‬‬‫ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﻋدد ﻧﺎدرﺳﺗﯽ‬‫را ﻧﺷﺎن دھد‪ .‬ﺑﺎر دﯾﮕر‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫در طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫زﻣﺎﻧﯽ آﺷﮑﺎر ﻣﯽﺷود ﮐﮫ در ﺣﯾن اﻧدازهﮔﯾری ﺑﺎزوﺑﻧد‬‫ﺷل ﺑﺎﺷد‪.‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﻋدد ﻧﺎدرﺳﺗﯽ‬‫را ﻧﺷﺎن دھد‪ .‬ﺑﺎزوﺑﻧد را‬‫ﺑﮫدرﺳﺗﯽ ﺑﺑﻧدﯾد‪ ،‬و ﺑﺎر دﯾﮕر‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫اﻧدازهﮔﯾریھﺎی ﻗﺑﻠﯽ ﮐﮫ در ﺣﺎﻓظﮫ ذﺧﯾره ﺷده اﺳت‪.‬‬‫ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت‬‫‪) FULL BATTERY‬ﺑﺎﺗری ﭘر اﺳت(‬‫ﻧﺷﺎﻧﮕر ﺗوان ﺑﺎﺗری در طول اﻧدازهﮔﯾری‪.‬‬‫‪) LOW BATTERY‬ﺑﺎﺗری ﮐم(‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﺎﺗری ﺿﻌﯾف ﺷود‪ ،‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﻓﺷﺎر ﺧون ﺑﮫ دﻟﯾل ﺗﮑﺎن ﺧوردن در طول اﻧدازهﮔﯾری‬‫ﻧﺎﭘﺎﯾدار اﺳت‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ اﯾن ﻧﺷﺎﻧﮫ ﭼﺷﻣﮏ‬‫ﻣﯽزﻧد‪ ،‬ھﻣﮫ ﺑﺎﺗریھﺎ را ﺗﻌوﯾض‬‫ﮐﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫در طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت‬‫ﺑﻣﺎﻧﯾد‪.‬‬‫ﻣﻘﺎدﯾر ﺳﯾﺳﺗوﻟﯽ و دﯾﺎﺳﺗوﻟﯽ ﮐﻣﺗر از ‪ mmHg 10‬ﺑﺎ‬‫ﯾﮑدﯾﮕر اﺧﺗﻼف دارﻧد‪.‬‬‫ﻣﻘدار ﻓﺷﺎر در زﻣﺎن ﺑﺎد ﺷدن اﻓزاﯾش ﻧﯾﺎﻓت‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫدرﺳﺗﯽ ﻗرار ﻧﮕرﻓﺗﮫ اﺳت‪.‬‬‫‪PUL DISPLAY ERROR‬‬‫ﻧﺑض ﺑﮫدرﺳﺗﯽ ﺷﻧﺎﺳﺎﯾﯽ ﻧﺷده اﺳت‪.‬‬‫ﺧطﺎی داﺧﻠﯽ دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون‬‫‪SYS‬‬‫ﻓﺷﺎر ﺧون ﺳﯾﺳﺗوﻟﯽ ﺑر ﺣﺳب ‪mmHg‬‬‫‪DIA‬‬‫ﻓﺷﺎر ﺧون دﯾﺎﺳﺗوﻟﯽ ﺑر ﺣﺳب ‪mmHg‬‬‫‪PUL‬‬‫ﻧﺑض در دﻗﯾﻘﮫ‬‫‪AM‬‬‫اطﻼﻋﺎﺗﯽ ﮐﮫ ﺑﯾن ﺳﺎﻋت ‪ 4:00‬ﺗﺎ ‪ 9:59‬ﮔرﻓﺗﮫ ﺷده اﺳت‬‫‪PM‬‬‫اطﻼﻋﺎﺗﯽ ﮐﮫ ﺑﯾن ﺳﺎﻋت ‪ 18:00‬ﺗﺎ ‪ 1:59‬ﮔرﻓﺗﮫ ﺷده‬‫اﺳت‬‫ﺗﻧظﯾﻣﺎت ﻓﺷﺎر‬‫ﻣﻘدار ﻓﺷﺎری را ﻧﺷﺎن ﻣﯽدھد ﮐﮫ ﻗﺑﻼً ﺗوﺳط ﮐﺎرﺑر ﺗﻧظﯾم‬‫ﺷده اﺳت‪.‬‬‫ﻓﺎرﺳﯽ ‪6‬‬‫ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ ﺑﺑﻧدﯾد‪ ،‬و‬‫اﻧدازهﮔﯾری را ﺗﮑرار ﮐﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ را ﺧﺎرج ﮐرده و دﮐﻣﮫ‬‫‪ START‬را ﺑزﻧﯾد‪ ،‬و ﺑﺎر دﯾﮕر‬‫ﺑﺎﺗریھﺎ را ﻧﺻب ﮐﻧﯾد‪ .‬اﮔر ﺧطﺎ‬‫ھﻣﭼﻧﺎن ظﺎھر ﺷد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﺗﻣﺎس‬‫ﺑﮕﯾرﯾد‪.‬‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت‬‫‪ .1‬اﻧدازهﮔﯾری ﻋﺎدی‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪ .‬ﻓﺷﺎر ﺧون اﻧدازهﮔﯾری ﺷده و اطﻼﻋﺎت درون ﺣﺎﻓظﮫ دﺳﺗﮕﺎه ذﺧﯾره‬‫ﻣﯽﺷود‪ .‬اﯾن دﺳﺗﮕﺎه ﻣﯽﺗواﻧد ﺑﮫ طور ﺧودﮐﺎر‪ ،‬ﻧود اﻧدازهﮔﯾری آﺧر را در ﺣﺎﻓظﮫاش ذﺧﯾره ﮐﻧد‪.‬‬‫‪ .2‬ﻓراﺧواﻧﯽ اطﻼﻋﺎت‬‫ﺟﮭت ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ‪ ،‬دﮐﻣﮫ ﯾﺎ را ﻓﺷﺎر دھﯾد‪.‬‬‫ھﻣﺎﻧﮕوﻧﮫ ﮐﮫ در ﺷﮑل ﺳﻣت راﺳت ﻧﺷﺎن داده ﺷده‪ ،‬ﻣﯾﺎﻧﮕﯾن‬‫ﺗﻣﺎم اﻧدازهﮔﯾریھﺎ ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫در اﯾن ﺣﺎﻟت‪ ،‬ﺑﺎ ھر ﺑﺎر ﻓﺷردن دﮐﻣﮫ‬‫ﺷﮑل زﯾر ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫اطﻼﻋﺎت ﺣﺎﻓظﮫ ﺑﮫ‬‫ﻣﯾﺎﻧﮕﯾن ﺗﻣﺎم اﻧدازهﮔﯾریھﺎی ‪) AM‬ﺻﺑﺢ( ﮐﮫ ﺑﯾن ﺳﺎﻋﺎت‬‫‪ 4:00‬ﺗﺎ ‪ 9:59‬اﻧﺟﺎم ﺷده اﺳت‪.‬‬‫ﻣﯾﺎﻧﮕﯾن ﺗﻣﺎم اﻧدازهﮔﯾریھﺎی ‪) PM‬ﺷب( ﮐﮫ ﺑﯾن ﺳﺎﻋﺎت‬‫‪ 18:00‬ﺗﺎ ‪ 1:59‬اﻧﺟﺎم ﺷده اﺳت‪.‬‬‫ﺟدﯾدﺗرﯾن اطﻼﻋﺎت )‪ No.n‬در ﻣﺛﺎل ﻧﺷﺎن داده ﺷده ‪(No.35‬‬‫ﻗدﯾﻣﯽﺗرﯾن اطﻼﻋﺎت )‪(No.1‬‬‫ﺑرای ﻣﺷﺎھده ﺟزﺋﯾﺎت ﺑﯾﺷﺗر در ﻣورد ﻓراﺧواﻧﯽ اطﻼﻋﺎت‪ ،‬ﺑﮫ‬‫ﺑﺧش ﻓراﺧواﻧﯽ "اطﻼﻋﺎت ﺣﺎﻓظﮫ ﻣراﺟﻌﮫ ﮐﻧﯾد"‪) .‬ﺑﮫ ﺻﻔﺣﮫ ‪18‬‬‫ﻣراﺟﻌﮫ ﮐﻧﯾد(‬‫‪ .3‬ﺣذف ﮐﻠﯾﮫ اطﻼﻋﺎت ذﺧﯾرهﺷده در ﺣﺎﻓظﮫ‬‫و‬‫دﮐﻣﮫھﺎی و را ﺑﮫ طور ھﻣزﻣﺎن ﻓﺷﺎر دھﯾد‪ .‬ﻋﻼﻣت‬‫ﻧﺷﺎﻧﮕر ﺑﺎﺗری ظﺎھر ﻣﯽﺷوﻧد‪ .‬ﺟﮭت ﺣذف ﺗﻣﺎم اطﻼﻋﺎت ذﺧﯾرهﺷده در‬‫ﺣﺎﻓظﮫ‪ ،‬دﮐﻣﮫھﺎی و را ﻓﺷﺎر داده و ﻧﮕﮫ دارﯾد ﺗﺎ زﻣﺎﻧﯽ ﮐﮫ‬‫ﻋﻼﻣت‬‫ﻧﻣﺎﯾش داده ﺷده ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐﻧد‪.‬‬‫‪ .4‬اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫ﺑﮫ ﺻﻔﺣﮫ ‪ 15‬ﻣراﺟﻌﮫ ﮐﻧﯾد ﺗﺎ اطﻼﻋﺎت ﺑﯾﺷﺗری در ﻣورد اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب ﺑﯾﺎﺑﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪7‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﺻب ‪ /‬ﺗﻌوﯾض ﺑﺎﺗریھﺎ‬‫‪ .1‬درﭘوش ﺑﺎطری را ﺑردارﯾد‪.‬‬‫‪ .2‬ﺑﺎﺗریھﺎی ﮐﮭﻧﮫ را ﺑردارﯾد و ھﻣﺎنطور ﮐﮫ ﻧﺷﺎن‬‫داده ﺷده اﺳت‪ ،‬ﺑﺎﺗریھﺎی ﻧو را در ﻣﺣﻔظﮫ‬‫ﺑﺎطری ﻗرار دھﯾد‪ .‬ﻣراﻗب ﺑﺎﺷﯾد ﮐﮫ ﻗطبھﺎی‬‫ﺑﺎﺗری )‪ +‬و ‪ (-‬در ﺟﺎی ﺻﺣﯾﺢ ﻗرار ﮔﯾرﻧد‪.‬‬‫ﺻرﻓﺎ ً از ﺑﺎﺗریھﺎی ‪ LR6 ،R6P‬ﯾﺎ ‪AA‬‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫‪ .3‬درﭘوش ﺑﺎﺗری را ﺑﮕذارﯾد‪.‬‬‫ﻣرﺣﻠﮫ ‪1‬‬‫ﻣرﺣﻠﮫ ‪3‬‬‫ﻣرﺣﻠﮫ ‪1‬‬‫ﻣرﺣﻠﮫ ‪2‬‬‫اﺣﺗﯾﺎط‬‫ﺑﺎﺗریھﺎ را طﺑﻖ ﺗﺻوﯾر در ﻣﺣﻔظﮫ ﺑﺎطری ﻗرار دھﯾد‪ .‬اﮔر ﺑﺎﺗریھﺎ ﺑﮫ ﺷﮑل ﻧﺎدرﺳت ﻗرار ﮔﯾرﻧد‪ ،‬دﺳﺗﮕﺎه‬‫ﮐﺎر ﻧﺧواھد ﮐرد‪.‬‬‫)ﻋﻼﻣت ‪ (LOW BATTERY‬ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐرد‪ ،‬ھﻣﮫ ﺑﺎﺗریھﺎ را ﺑﺎ ﺑﺎﺗریھﺎی‬‫وﻗﺗﯽ‬‫ﻧو ﺗﻌوﯾض ﮐﻧﯾد‪ .‬ﺑﺎﺗریھﺎی ﮐﮭﻧﮫ و ﻧو را ﺑﺎ ھم در دﺳﺗﮕﺎه ﻗرار ﻧدھﯾد‪ .‬اﯾن ﮐﺎر ﻣﯽﺗواﻧد ﻣوﺟب ﮐوﺗﺎه ﺷدن‬‫ﻋﻣر ﺑﺎﺗری ﯾﺎ ﻧﻘص در ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﺑﺎﺗریھﺎ را دو ﯾﺎ ﭼﻧد ﺛﺎﻧﯾﮫ ﺑﻌد از ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه ﺗﻌوﯾض ﮐﻧﯾد‪.‬‬‫اﮔر‬‫)ﻋﻼﻣت ‪ (LOW BATTERY‬ﺣﺗﯽ ﭘس از ﺗﻌوﯾض ﺑﺎﺗریھﺎ ظﺎھر ﺷد‪ ،‬ﯾﮏﺑﺎر اﻧدازهﮔﯾری‬‫ﻓﺷﺎر ﺧون را اﻧﺟﺎم دھﯾد‪ .‬آﻧﮕﺎه دﺳﺗﮕﺎه ﻣﯽﺗواﻧد ﺑﺎﺗریھﺎی ﺟدﯾد را ﺷﻧﺎﺳﺎﯾﯽ ﮐﻧد‪.‬‬‫)ﻋﻼﻣت ‪ (LOW BATTERY‬ظﺎھر ﻧﻣﯽﺷود‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﺎﺗریھﺎ ﺧﺎﻟﯽ ﺷده ﺑﺎﺷﻧد‪،‬‬‫ﻋﻣر ﺑﺎﺗری ﺑﺳﺗﮫ ﺑﮫ دﻣﺎی ﭘﯾراﻣوﻧﯽ ﻣﯽﺗواﻧد ﻣﺗﻐﯾر ﺑﺎﺷد و در دﻣﺎی ﮐم ﻣﻣﮑن اﺳت ﻋﻣر ﺑﺎﺗری ﮐﻣﺗر ﺷود‪.‬‬‫ﻣﻌﻣوﻻً‪ ،‬ﭼﮭﺎر ﺑﺎﺗری ﻧوی ‪ LR6‬در دﺳﺗﮕﺎھﯽ ﮐﮫ دو ﺑﺎر در روز اﺳﺗﻔﺎده ﻣﯽﺷود‪ ،‬ﺣدودا ً ﯾﮏ ﺳﺎل ﻋﻣر‬‫ﻣﯽﮐﻧد‪.‬‬‫ﺻرﻓﺎ ً از ﺑﺎﺗریھﺎی ﺗﻌﯾﯾنﺷده اﺳﺗﻔﺎده ﮐﻧﯾد‪ .‬ﺑﺎﺗریھﺎﯾﯽ ﮐﮫ ھﻣراه ﺑﺎ دﺳﺗﮕﺎه ﺑﮫ ﺷﻣﺎ داده ﻣﯽﺷوﻧد‪ ،‬ﺑرای‬‫آزﻣﺎﯾش ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ھﺳﺗﻧد و ﻟذا ﻣﻣﮑن اﺳت ﻋﻣر ﻣﺣدودی داﺷﺗﮫ ﺑﺎﺷﻧد‪.‬‬‫اﮔر ﻗرار اﺳت ﮐﮫ از دﺳﺗﮕﺎه ﺑرای ﻣدت طوﻻﻧﯽ اﺳﺗﻔﺎده ﻧﮑﻧﯾد‪ ،‬ﺑﺎﺗریھﺎﯾش را ﺧﺎرج ﮐﻧﯾد‪.‬‬‫ﻣﻣﮑن اﺳت ﺑﺎﺗری ﻧﺷت ﮐﻧد و ﺑﺎﻋث اﺧﺗﻼل در ﮐﺎرﮐرد دﺳﺗﮕﺎه ﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪8‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻣﺗﺻل ﮐردن ﺷﯾﻠﻧﮓ ھوا‬‫ﻧری راﺑط ھوا را ﺑﮫ داﺧل ﻣﺎدﮔﯽ ھوا وارد ﮐرده‬‫و از ﻣﺣﮑم ﺑودن آن اطﻣﯾﻧﺎن ﺣﺎﺻل ﮐﻧﯾد‪.‬‬‫ﻣﺎدﮔﯽ ھوا‬‫ﮔ‬‫ﻧری راﺑط ھوا‬‫ﻣﺗﺻل ﮐردن آداﭘﺗور ‪AC‬‬‫ﻧری آداﭘﺗور ‪ AC‬را ﺑﮫ ﻓﯾش ‪ DC‬وارد ﮐﻧﯾد‪.‬‬‫ﺳﭘس‪ ،‬آداﭘﺗور ‪ AC‬را ﺑﮫ ﭘرﯾز ﺑرق ﻣﺗﺻل ﮐﻧﯾد‪.‬‬‫ﻧری آداﭘﺗور ‪AC‬‬‫از آداﭘﺗور ‪ AC‬ﻣﺷﺧصﺷده اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫)ﺑﮫ ﺻﻔﺣﮫ ‪ 23‬ﻣراﺟﻌﮫ ﮐﻧﯾد‪(.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد آداﭘﺗور ‪ AC‬را از ﭘرﯾز ﺑرق ﺟدا ﮐﻧﯾد‪ ،‬ﺑدﻧﮫ آداﭘﺗور ‪ AC‬را ﺑﮕﯾرﯾد و ﺑﮑﺷﯾد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﻣﯽﺧواھﯾد ﻓﯾش آداﭘﺗور ‪ AC‬را از دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺟدا ﮐﻧﯾد‪ ،‬ﻓﯾش آداﭘﺗور ‪ AC‬را‬‫ﮔرﻓﺗﮫ و از دﺳﺗﮕﺎه ﺑﯾرون ﺑﮑﺷﯾد‪.‬‬‫ﻓﯾش ‪DC‬‬‫ﻓﺎرﺳﯽ ‪9‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﺗﻧظﯾم ﺳﺎﻋت داﺧﻠﯽ‬‫را ﻓﺷﺎر دھﯾد‬‫ﻗﺑل از اﺳﺗﻔﺎده‪ ،‬ﺳﺎﻋت را ﺗﻧظﯾم ﮐﻧﯾد‪.‬‬‫‪ .1‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر دھﯾد ﺗﺎ زﻣﺎﻧﯽ ﮐﮫ ﻋدد ﻣرﺑوط ﺑﮫ ﺳﺎل‬‫ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐﻧد‪.‬‬‫ﯾﺎ‬‫‪ .2‬ﺑﺎ اﺳﺗﻔﺎده از دﮐﻣﮫھﺎی ﯾﺎ ﺳﺎل ﻣورد ﻧظر را اﻧﺗﺧﺎب ﮐﻧﯾد‪.‬‬‫ﺟﮭت ﺗﻧظﯾم ﺳﺎل و اﻧﺗﺧﺎب ﻣﺎه‪/‬روز‪ ،‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر‬‫دھﯾد‪ .‬ﻣﺣدوده ﺳﺎلھﺎی ﻗﺎﺑل اﻧﺗﺧﺎب ﺟﮭت ﺗﻧظﯾم ﺗﺎرﯾﺦ‪ ،‬از‬‫‪ 2010‬ﺗﺎ ‪ 2059‬ﻣﯽﺑﺎﺷد‪.‬‬

‫ﺳﺎل‬

‫ﯾﺎ‬

‫ﻣﺎه‬

‫‪ .3‬ﺑﺎ اﺳﺗﻔﺎده از دﮐﻣﮫھﺎی ﯾﺎ ﻣﺎه ﻣورد ﻧظر را اﻧﺗﺧﺎب ﮐﻧﯾد‪.‬‬‫ﺟﮭت ﺗﻧظﯾم ﻣﺎه و اﻧﺗﺧﺎب روز‪ ،‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر دھﯾد‪.‬‬‫‪ .4‬ﺑﺎ اﺳﺗﻔﺎده از ﮐﻠﯾدھﺎی ﯾﺎ روز ﻣورد ﻧظر را اﻧﺗﺧﺎب ﮐﻧﯾد‪.‬‬‫ﺟﮭت ﺗﻧظﯾم روز و اﻧﺗﺧﺎب ﺳﺎﻋت‪/‬دﻗﯾﻘﮫ‪ ،‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر‬‫دھﯾد‪.‬‬‫ﯾﺎ‬‫‪ .5‬ﺑﺎ اﺳﺗﻔﺎده از دﮐﻣﮫھﺎی ﯾﺎ ﺳﺎﻋت ﻣورد ﻧظر را اﻧﺗﺧﺎب ﮐﻧﯾد‪.‬‬‫ﺟﮭت ﺗﻧظﯾم ﺳﺎﻋت و اﻧﺗﺧﺎب دﻗﯾﻘﮫ‪ ،‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر‬‫دھﯾد‪.‬‬‫‪ .6‬ﺑﺎ اﺳﺗﻔﺎده از دﮐﻣﮫھﺎی ﯾﺎ دﻗﯾﻘﮫ ﻣورد ﻧظر را اﻧﺗﺧﺎب ﮐﻧﯾد‪.‬‬‫)ﺟﮭت رﻓﺗن ﺑﮫ ﺣﺎﻟت ﺗﻧظﯾم ﻓﺷﺎر‪ ،‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ﺟﮭت ﮐﺳب اطﻼﻋﺎت ﺑﯾﺷﺗر ﺑﮫ ﺻﻔﺣﮫ ‪ 15‬ﻣراﺟﻌﮫ ﮐﻧﯾد‪(.‬‬‫ﺑرای ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه‪ ،‬دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ﺗوﺟﮫ‪ :‬ﭘس از ﺳﮫ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺧودش ﺑﮫطور ﺧودﮐﺎر‬‫ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺳﺎﻋت ﺗﻧظﯾم ﻧﺷده اﺳت‪،‬‬‫ھﻣﺎﻧﮕوﻧﮫ ﮐﮫ در ﺳﻣت راﺳت ﻧﺷﺎن داده‬‫ﺷده‪ ،‬ﻧﺷﺎﻧﮕر ﺳﺎﻋت ﺑﮫ ﺟﺎی اﻋداد‪ ،‬ﺧط‬‫ﺗﯾره ﻧﻣﺎﯾش ﻣﯽدھد‪.‬‬‫ﺑﺎ ﻧﮕﮫ داﺷﺗن ﮐﻠﯾدھﺎی ﯾﺎ ﻣﯽﺗواﻧﯾد ﻣﻘﺎدﯾر‬‫را ﺑﮫ طور ﭘﯾوﺳﺗﮫ ﺗﻐﯾﯾر دھﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪10‬‬‫روز‬‫ﺳﺎﻋت‬‫ﯾﺎ‬‫ﯾﺎ‬‫دﻗﯾﻘﮫ‬‫ﺧﺎﻣوش ﮐردن‬‫ﻓﺷردن دﮐﻣﮫ‬‫ﺣﺎﻟت ‪ START‬در ھر‬‫ﺗﻧظﯾم ﻓﺷﺎر ﻟﺣظﮫ ﻣوﺟب ﺧﺎﻣوش‬‫ﺷدن دﺳﺗﮕﺎه ﺧواھد ﺷد‪.‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫اﻧﺗﺧﺎب ﺳﺎﯾز ﻣﻧﺎﺳب ﺑﺎزوﺑﻧد‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧدی ﺑﺎ اﻧدازه ﻣﻧﺎﺳب ﻧﻘش ﻣﮭﻣﯽ در ﻗراﺋت ﺻﺣﯾﺢ دﺳﺗﮕﺎه دارد‪ .‬اﮔر اﻧدازه ﺑﺎزوﺑﻧد ﻣﻧﺎﺳب‬‫ﻧﺑﺎﺷد‪ ،‬ﻣﻣﮑن اﺳت ﻗراﺋت دﺳﺗﮕﺎه دﭼﺎر اﺷﺗﺑﺎه ﺷود‪.‬‬‫اﻧدازه ﺑﺎزو روی ھر ﺑﺎزوﺑﻧد ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺷﺎﺧص و داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب روی ﺑﺎزوﺑﻧد ﺑﮫ ﺷﻣﺎ ﻣﯽﮔوﯾد ﮐﮫ آﯾﺎ از ﺑﺎزوﺑﻧد ﺻﺣﯾﺣﯽ اﺳﺗﻔﺎده ﻣﯽﮐﻧﯾد‬‫ﯾﺎ ﺧﯾر‪) .‬ﺑﮫ ﺑﺧش "ﻧﻣﺎدھﺎی ﭼﺎپﺷده روی ﺑﺎزوﺑﻧد" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد(‬‫اﮔر ﺷﺎﺧص ﺧﺎرج از داﻣﻧﮫ ﻣورد ﻧظر اﺳت‪ ،‬ﺟﮭت ﺧرﯾد ﺑﺎزوﺑﻧد دﯾﮕر ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽﺗﺎن ﺗﻣﺎس‬‫ﺑﮕﯾرﯾد‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫﻋﻧوان ﯾﮏ ﮐﺎﻻی ﻣﺻرﻓﯽ ﻗﻠﻣداد ﻣﯽﺷود‪ .‬اﮔر ﺑﺎزوﺑﻧد ﭘﺎره ﺷد‪ ،‬ﯾﮑﯽ دﯾﮕر ﺑﺧرﯾد‪.‬‬‫اﻧدازه ﺗوﺻﯾﮫ ﺷده ﺑﺎزوﺑﻧد‬‫اﻧدازه ﺑﺎزو‬‫‪ 31‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 45‬ﺳﺎﻧﺗﯽﻣﺗر ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 42‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 32‬ﺳﺎﻧﺗﯽﻣﺗر‬‫‪ 16‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 24‬ﺳﺎﻧﺗﯽﻣﺗر ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ ﺑزرﮔﺳﺎﻻن‬‫اﻧدازه ﺑﺎزو‪ :‬ﻣﺣﯾط ﺑﺎزو در ﺑﺧش ﻋﺿﻠﮫ دو ﺳر‪.‬‬‫اﺳﺗﻔﺎده از ﺑﺎزوﺑﻧد‬‫‪ .1‬ھﻣﺎنطور ﮐﮫ در ﺗﺻوﯾر ﻣﺷﺎھده ﻣﯽﮐﻧﯾد‪ ،‬ﺑﺎزوﺑﻧد‬‫را دور ﺑﺎزو‪ ،‬ﯾﻌﻧﯽ ‪ 1‬اﻟﯽ ‪ 2‬ﺳﺎﻧﺗﯽﻣﺗر ﺑﺎﻻﺗر از‬‫ﮔودی آرﻧﺞ ﺑﺑﻧدﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را ﻣﺳﺗﻘﯾﻣﺎ ً روی ﭘوﺳت ﻗرار دھﯾد‪ ،‬زﯾرا‬‫ﻟﺑﺎس ﻣﯽﺗواﻧد ﺑﺎﻋث ﺗﺿﻌﯾف ﺿرﺑﺎن ﺷود و در‬‫ﻗراﺋت ﺧطﺎ اﯾﺟﺎد ﮐﻧد‪.‬‬‫‪ .2‬ﺗﻧﮓ ﺷدن ﺑﺎزو ﺑر اﺛر ﺑﺎﻻ زدن زﯾﺎد آﺳﺗﯾن‬‫ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺧطﺎ در ﻗراﺋت دﺳﺗﮕﺎه ﺷود‪.‬‬‫‪ .3‬ﻣطﻣﺋن ﺷوﯾد ﮐﮫ ﺷﺎﺧص‬‫ﻣﻧﺎﺳﺑﯽ ﻗرار دارد‪.‬‬‫آﺳﺗﯾنھﺎ را زﯾﺎد ﺑﺎﻻ ﻧزﻧﯾد‪.‬‬‫‪ 2 - 1‬ﺳﺎﻧﺗﯽﻣﺗر‬‫ﺑﺎزوﺑﻧد‬‫ﺷﯾﻠﻧﮓ ھوا‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫ﭼﺳب ﺑﺎزوﺑﻧد را‬‫ﻣﺣﮑم ﮐﻧﯾد‪.‬‬‫در داﻣﻧﮫ اﻧدازه‬‫ﺷﺎﺧص‬‫ﻓﺎرﺳﯽ ‪11‬‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫‪CUF-F-SA‬‬‫و داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﻣﺎدھﺎی ﭼﺎپ ﺷده روی ﺑﺎزوﺑﻧد‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﻧﺷﺎﻧﮫ ﻣوﻗﻌﯾت ﺷرﯾﺎن‬‫▲‬‫‪REF‬‬‫ﺷﺎﺧص‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‪.‬‬‫اﯾن ﻋﻼﻣت روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫را روی ﺷرﯾﺎن ﺑﺎزو ﯾﺎ‬‫ﻧﺷﺎﻧﮫ‬‫ھمراﺳﺗﺎ ﺑﺎ اﻧﮕﺷت ﺣﻠﻘﮫ در داﺧل‬‫ﺑﺎزو ﺗﻧظﯾم ﮐﻧﯾد‪.‬‬‫از ﺑﺎزوﺑﻧد ﺑزرگ ﺑزرﮔﺳﺎﻻن ﺑﮫﺟﺎی‬‫ﺧﺎرج از داﻣﻧﮫای ﮐﮫ روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎل و‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﯾﺎ ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﻋرﯾض ﭼﺎپ ﻣﯽﺷود‪.‬‬‫ﻋرﯾض اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﻋرﯾض‪ .‬روی‬‫ﺑﺎزوﺑﻧد ﻋرﯾض ﭼﺎپ ﺷده اﺳت‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ‬‫ﺑزرﮔﺳﺎﻻن‪ .‬اﯾن ﻋﻼﻣت روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ‬‫ﺑزرﮔﺳﺎل ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺧﺎرج از داﻣﻧﮫای ﮐﮫ روی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎل و از ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ ﺑزرﮔﺳﺎﻻن‬‫ﺑﮫﺟﺎی ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﯾﺎ ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﻋرﯾض ﭼﺎپ ﻣﯽﺷود‪.‬‬‫ﻋرﯾض اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب ﺑرای ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ‬‫ﺑزرﮔﺳﺎﻻن‪ .‬اﯾن ﻋﻼﻣت روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ‬‫ﺑزرﮔﺳﺎﻻن ﭼﺎپ ﺷده اﺳت‪.‬‬‫از ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن ﺑﮫﺟﺎی‬‫اﯾن ﻋﻼﻣت روی ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن ﯾﺎ‬‫ﭼﺎپ ﺷده اﺳت‪.‬‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ ﺑزرﮔﺳﺎﻻن‬‫اﺳﺗﻔﺎده ﮐﻧﯾد‪.‬‬‫ﺷﻣﺎره دﺳﺗﮫ‬‫داﻣﻧﮫ اﻧدازه ﻣﻧﺎﺳب‬
‫‪L‬‬

‫‪A‬‬

‫‪L‬‬

‫‪W‬‬

‫‪S‬‬

‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ‬‫ﺑزرﮔﺳﺎﻻن‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬

‫‪L‬‬

‫‪A‬‬

‫‪S‬‬

‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬

‫‪A‬‬

‫‪S‬‬

‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ ﺑزرﮔﺳﺎﻻن‬‫ﻓﺎرﺳﯽ ‪12‬‬‫اﺳﺗﻔﺎده از دﺳﺗﮕﺎه ﭘﺎﯾش‬‫ﻧﺣوه اﻧﺟﺎم اﻧدازهﮔﯾری دﻗﯾﻖ‬‫ﺑﮫﻣﻧظور اﻧﺟﺎم دﻗﯾﻖﺗرﯾن اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪:‬‬‫آرام روی ﺻﻧدﻟﯽ ﺑﻧﺷﯾﻧﯾد‪ .‬دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﺑﮕذارﯾد‪ .‬ﭘﺎھﺎﯾﺗﺎن را رویھم ﻧﯾﻧدازﯾد‪ .‬ﭘﺎھﺎﯾﺗﺎن را‬‫ﺻﺎف روی زﻣﯾن ﺑﮕذارﯾد و ﮐﻣرﺗﺎن را ﺻﺎف ﮐﻧﯾد‪.‬‬‫ﺑﮫ ﻣدت ﭘﻧﺞ اﻟﯽ ده دﻗﯾﻘﮫ آرام ﺑﺎﺷﯾد و ﺳﭘس اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫ﻣرﮐز ﺑﺎزوﺑﻧد را ھمﺗراز ﺑﺎ ﻗﻠﺑﺗﺎن ﻗرار دھﯾد‪.‬‬‫در طول اﻧدازهﮔﯾری ﺑﯽﺣرﮐت و آرام ﺑﻣﺎﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻧﺑﺎﯾد ﺑﻼﻓﺎﺻﻠﮫ ﭘس از ورزش ﯾﺎ ﺣﻣﺎم ﮐردن اﻧﺟﺎم ﺷود‪ .‬ﺑﯾﺳت اﻟﯽ ﺳﯽ دﻗﯾﻘﮫ آرام ﺑﻣﺎﻧﯾد و ﺑﻌد‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫ﺳﻌﯽ ﮐﻧﯾد اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون را ھر روز در ﯾﮏ ﺳﺎﻋت ﻣﻌﯾن اﻧﺟﺎم دھﯾد‪.‬‬‫اﻧدازهﮔﯾری‬‫در طول اﻧدازهﮔﯾری‪ ،‬طﺑﯾﻌﯽ اﺳت ﮐﮫ اﺣﺳﺎس ﮐﻧﯾد ﺑﺎزوﺑﻧد ﺧﯾﻠﯽ ﻣﺣﮑم ﺑﺳﺗﮫ ﺷده اﺳت‪) .‬ﻧﮕران ﻧﺑﺎﺷﯾد(‪.‬‬‫ﭘس از اﻧدازهﮔﯾری‬‫ﭘس از اﻧدازهﮔﯾری‪ ،‬دﮐﻣﮫ ‪ START‬را ﺑزﻧﯾد ﺗﺎ دﺳﺗﮕﺎه ﺧﺎﻣوش ﺷود‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑﺎز ﮐﻧﯾد و اطﻼﻋﺎت را ﺛﺑت ﮐﻧﯾد‪ .‬ﭘس از ﯾﮏ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺧودش ﺑﮫطور‬‫ﺧودﮐﺎر ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫ﻓﺎرﺳﯽ ‪13‬‬‫اﻧدازهﮔﯾریھﺎ‬‫ﭘﯾش از اﻧدازهﮔﯾری‪ ،‬ﺑﺧش "ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ" در ﺻﻔﺣﮫ ‪ 16‬را ﻣطﺎﻟﻌﮫ ﮐﻧﯾد‪.‬‬‫اﻧدازهﮔﯾری ﻋﺎدی‬‫‪.1‬‬‫ﺑﺎزوﺑﻧد را دور ﺑﺎزو )ﺗرﺟﯾﺣﺎ ً روی ﺑﺎزوی ﭼپ( ﻗرار دھﯾد‪.‬‬‫در ﺣﯾن اﻧدازهﮔﯾری آرام ﺑﻣﺎﻧﯾد‪.‬‬‫‪.2‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ھﻣﮫ ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر ﻧﻣﺎﯾﺎن ﻣﯽﺷوﻧد‪.‬‬‫ﺻﻔر )‪ (0‬ﻧﻣﺎﯾﺎن ﻣﯽﺷود ﮐﮫ اﻧدﮐﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ اﻧدازهﮔﯾری آﻏﺎز ﻣﯽﺷود‪ ،‬ھﻣﺎنطور ﮐﮫ در‬‫ﺗﺻوﯾر ﺳﻣت راﺳت ﻧﺷﺎن داده ﺷده اﺳت‪ ،‬ﻧﻣﺎﯾﺷﮕر ﺗﻐﯾﯾر‬‫ﻣﯽﮐﻧد‪ .‬ﺑﺎزوﺑﻧد ﺷروع ﺑﮫ ﺑﺎد ﺷدن ﻣﯽﮐﻧد‪ .‬اﺣﺳﺎس ﺗﻧﮕﯽ‬‫ﺑﺎزوﺑﻧد طﺑﯾﻌﯽ اﺳت‪ .‬ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر در ﻟﺑﮫ ﺳﻣت ﭼپ‬‫ﻧﻣﺎﯾﺷﮕر در طول ﺑﺎد ﮐردن ﻧﻣﺎﯾﺎن ﻣﯽﺷود‪.‬‬‫ﻧﮑﺗﮫ‪ :‬ھر زﻣﺎن ﺧواﺳﺗﯾد ﺑﺎد ﮐردن را ﻣﺗوﻗف ﮐﻧﯾد‪ ،‬دﮐﻣﮫ‬‫‪ START‬را ﻣﺟددا ً ﻓﺷﺎر دھﯾد‪.‬‬‫‪.3‬‬‫وﻗﺗﯽ ﺑﺎد ﮐردن ﺗﻣﺎم ﺷد‪ ،‬ﺧﺎرج ﺷدن ﺑﺎد ﺑﮫﺻورت ﺧودﮐﺎر‬‫آﻏﺎز ﺷده و‬‫)ﻋﻼﻣت ﻗﻠب( ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن‬‫ﻣﯽﮐﻧد‪ ،‬ﮐﮫ ﻧﺷﺎن ﻣﯽدھد اﻧدازهﮔﯾری در ﺣﺎل اﻧﺟﺎم اﺳت‪.‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺿرﺑﺎن ﻗﻠب ﺗﺷﺧﯾص داده ﺷد‪ ،‬ﻋﻼﻣت ﻗﻠب ﺑﺎ ھر‬‫ﺿرﺑﺎن ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫ﻧﮑﺗﮫ‪ :‬اﮔر ﻓﺷﺎر ﻣﻧﺎﺳب ﺑﮫ دﺳت ﻧﯾﺎﻣد‪ ،‬دﺳﺗﮕﺎه ﺑﮫطور ﺧودﮐﺎر‬‫ﻣﺟددا ً ﺷروع ﺑﮫ ﺑﺎد ﺷدن ﻣﯽﮐﻧد‪.‬‬‫ﺑرای اﺟﺗﻧﺎب از ﺑﺎد ﺷدن ﻣﺟدد‪ ،‬ﺑﮫ ﺑﺧش "اﻧدازهﮔﯾری ﺑﺎ‬‫ﻓﺷﺎر ﺗﻧظﯾم ﺷده" در ﺻﻔﺣﮫ ﺑﻌد ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫‪.4‬‬‫وﻗﺗﯽ اﻧدازهﮔﯾری ﺗﻣﺎم ﺷد‪ ،‬ﻗراﺋت ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ و دﯾﺎﺳﺗوﻟﯽ‬‫و ﺿرﺑﺎن ﻧﺑض ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‪.‬‬‫ھوای ﺑﺎﻗﯽﻣﺎﻧده از ﺑﺎزوﺑﻧد ﺧﺎرج ﺷده و ﺑﺎد آن ﺑﮫطور ﮐﺎﻣل‬‫ﺧﺎﻟﯽ ﻣﯽﺷود‪.‬‬‫‪.5‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد ﺗﺎ دﺳﺗﮕﺎه ﺧﺎﻣوش ﺷود‪.‬‬‫ﭘس از ﯾﮏ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺧودش ﺑﮫطور‬‫ﺧودﮐﺎر ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫ﻧﮑﺗﮫ‪:‬‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد را ﺑﺎ ﻓﺎﺻﻠﮫ ﺣداﻗل‬‫ﺳﮫ دﻗﯾﻘﮫ اﻧﺟﺎم دھﯾد‪.‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫‪ START‬را ﻓﺷﺎر دھﯾد‬‫ﮐﻠﯾﮫ ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر‬‫ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‬‫ﻧﻣﺎﯾش ﺻﻔر‬‫ﺷروع ﺑﺎد ﺷدن‬‫در ﺣﺎل اﻓزاﯾش ﻓﺷﺎر‬‫اﻧدازهﮔﯾری در ﺣﺎل‬‫اﻧﺟﺎم اﺳت‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‬‫طﺑﻘﮫﺑﻧدی ‪WHO‬‬‫ﺿرﺑﺎن ﻧﺑض‬‫ھوای ﺑﺎﻗﯽﻣﺎﻧده را ﺑﮫﺻورت‬‫ﺧودﮐﺎر ﺑﯾرون ﻣﯽدھد‬‫ﻓﺎرﺳﯽ ‪14‬‬‫اﻧدازهﮔﯾریھﺎ‬‫‪ UA-1020-W‬ﺑرای ﺷﻧﺎﺳﺎﯾﯽ ﺿرﺑﺎن ﻗﻠب و ﺑﺎد ﮐردن ﺧودﮐﺎر ﺑﺎزوﺑﻧد ﺗﺎ رﺳﯾدن ﺑﮫ ﺳطﺢ ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ ﻣﺷﺧص طراﺣﯽ ﺷده اﺳت‪.‬‬‫در ﺻورت ﺑﺎد ﺷدن ﻣﺟدد و ﻣﮑرر دﺳﺗﮕﺎه‪ ،‬روشھﺎی زﯾر را ﺑﮫ ﮐﺎر ﺑﺑرﯾد‪.‬‬‫اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺗﻧظﯾم ﺷده‬‫در طول ﻓراﯾﻧد اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون‪ ،‬ﻣﻣﮑن اﺳت ﺑﺎزوﺑﻧد ﻣﺟددا ً ﺑﺎد ﺷود‪.‬‬‫ﺟﮭت ﺟﻠوﮔﯾری از ﺑﺎد ﺷدن ﻣﺟدد‪ ،‬اﻣﮑﺎن ﺗﻧظﯾم ﯾﮏ ﻓﺷﺎر ﺛﺎﺑت وﺟود دارد‪.‬‬‫‪.1‬‬‫در ﻣرﺣﻠﮫ ‪ 6‬ﺗﻧظﯾم ﺳﺎﻋت )ﺻﻔﺣﮫ ‪ 10‬را ﻣﺷﺎھده ﮐﻧﯾد(‪ ،‬ﺟﮭت‬‫رﻓﺗن ﺑﮫ ﺣﺎﻟت ﺗﻧظﯾم ﻓﺷﺎر‪ ،‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر دھﯾد‪ .‬ﺣﺎﻟت‬‫ﺗﻧظﯾم ﺷده ﻓﻌﻠﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫‪.2‬‬‫ﺑﺎ ﻓﺷردن ﮐﻠﯾدھﺎی ﯾﺎ و از ﻣﯾﺎن ﺣﺎﻟتھﺎی زﯾر‪ ،‬ﻣﻘدار‬‫ﻓﺷﺎری ﮐﮫ ﺣدود ‪ mmHg 30‬ﯾﺎ ﺑﯾﺷﺗر از ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻣورد اﻧﺗظﺎر ﺷﻣﺎ ﺑﯾﺷﺗر اﺳت‪ ،‬اﻧﺗﺧﺎب ﮐﻧﯾد‪.‬‬‫‪ : AUTO‬ﺗﻧظﯾم ﻓﺷﺎر ﺧودﮐﺎر )ﻣﻘدار ﭘﯾشﻓرض(‬‫‪ : 180‬ﻣﻘدار ﻓﺷﺎر ‪) mmHg 180‬ﺛﺎﺑت(‬‫‪ : 210‬ﻣﻘدار ﻓﺷﺎر ‪) mmHg 210‬ﺛﺎﺑت(‬‫‪ : 240‬ﻣﻘدار ﻓﺷﺎر ‪) mmHg 240‬ﺛﺎﺑت(‬‫‪.3‬‬‫ﯾﺎ‬‫‪TriCheck‬‬‫ﺣﺎﻟت ﺗﻧظﯾم ﺧﺎﻣوش‬‫ﺟﮭت رﻓﺗن ﺑﮫ ﺣﺎﻟت ﺗﻧظﯾم ‪ ،TriCheck‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ﺑرای ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه‪ ،‬دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪ .‬ﭘس از ﺳﮫ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪،‬‬‫ﺧودش ﺑﮫطور ﺧودﮐﺎر ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫اﻧدازهﮔﯾری ﺑﻌدی ﺑﺎ ﻣﻘدار ﻓﺷﺎر ﺟدﯾد اﻧﺟﺎم ﺧواھدﺷد‪.‬‬‫اﻧدازهﮔﯾری ﺑﺎ ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫از اﯾن روش ھﻧﮕﺎﻣﯽ اﺳﺗﻔﺎده ﮐﻧﯾد ﮐﮫ ﺣﺗﯽ ﺑﺎ ﺗﻧظﯾم ﻓﺷﺎر روی ‪ 240‬ﺑﺎد ﮐردن ﻣﺟدد ﭼﻧدﯾن ﺑﺎر ﺗﮑرار‬‫ﻣﯽﺷود ﯾﺎ ھﻧﮕﺎﻣﯽ ﮐﮫ ﺣﺗﯽ ﺑﺎ ﮐﺎھش ﻓﺷﺎر ﺑﮫ ‪ mmHg 20‬ﯾﺎ ﮐﻣﺗر ﻧﺗﺎﯾﺞ ﻧﺷﺎن داده ﻧﻣﯽﺷوﻧد‪.‬‬‫‪.1‬‬‫ﺑﺎزوﺑﻧد را دور ﺑﺎزو )ﺗرﺟﯾﺣﺎ ً روی ﺑﺎزوی ﭼپ( ﻗرار دھﯾد‪.‬‬‫‪.2‬‬‫زﻣﺎﻧﯽ ﮐﮫ ﻋﻼﻣت ﺻﻔر ﺷروع ﺑﮫ ﭼﺷﻣﮏ زدن ﮐرد‪،‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد و ﻧﮕﮫدارﯾد ﺗﺎ ﻋددی در‬‫ﺣدود ‪ 30‬ﺗﺎ ‪ mmHg 40‬ﺑﺎﻻﺗر از ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‬‫ﻣورد اﻧﺗظﺎر ظﺎھر ﺷود‪.‬‬‫‪.3‬‬‫ھﻧﮕﺎﻣﯽ ﮐﮫ ﺑﮫ رﻗم ﻣورد ﻧظر رﺳﯾدﯾد‪ ،‬دﮐﻣﮫ‬‫‪ START‬را رھﺎ ﮐﻧﯾد ﺗﺎ اﻧدازهﮔﯾری آﻏﺎز ﺷود‪ .‬ﺑﮫ‬‫ھﻣﺎن ﺗرﺗﯾﺑﯽ ﮐﮫ در ﺻﻔﺣﮫ ﻗﺑل ﺗوﺿﯾﺢ داده ﺷد‪ ،‬ﺑﮫ‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧوﻧﺗﺎن اداﻣﮫ دھﯾد‪.‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫‪START‬‬‫دﮐﻣﮫ را ﻓﺷﺎر دھﯾد‬‫و ﻧﮕﮫدارﯾد‬‫دﮐﻣﮫ را در ﻓﺷﺎر‬‫ﺳﯾﺳﺗوﻟﯽ ﻣطﻠوب‬‫رھﺎ ﮐﻧﯾد‬‫ﺑرای ﻣﺷﺎھده ﺟزﺋﯾﺎت ﺑﯾﺷﺗر در ﻣورد‬‫اﻧدازهﮔﯾری ﺑﮫ ﺻﻔﺣﮫ ﻗﺑل ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫ﻓﺎرﺳﯽ ‪15‬‬‫ﻧﮑﺎﺗﯽ ﺑرای اﻧدازهﮔﯾری دﻗﯾﻖ‬‫در ﻣوﻗﻌﯾت راﺣﺗﯽ ﺑﻧﺷﯾﻧﯾد‪ .‬دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﻗرار دھﯾد ﺑﮫ ﮔوﻧﮫای ﮐﮫ ﮐف دﺳتھﺎﯾﺗﺎن رو ﺑﮫ‬‫ﺑﺎﻻ ﺑوده و ﺑﺎزوﺑﻧد و ﻗﻠﺑﺗﺎن ھمﺗراز ﺑﺎﺷﻧد‪.‬‬‫ﭘﯾش از آﻏﺎز اﻧدازهﮔﯾری‪ ،‬ﺣدود ﭘﻧﺞ ﺗﺎ ده دﻗﯾﻘﮫ آرام ﺑﺎﺷﯾد‪ .‬اﮔر ﺑﮫ دﻟﯾل ﺗﻧشھﺎی ھﯾﺟﺎﻧﯽ دﭼﺎر ھﯾﺟﺎن ﯾﺎ‬‫اﻓﺳردﮔﯽ ھﺳﺗﯾد‪ ،‬ﻓﺷﺎر ﺧوﻧﺗﺎن ﺑﺎﻻﺗر )ﯾﺎ ﭘﺎﯾﯾنﺗر( از ﻓﺷﺎر ﺧون ﻧرﻣﺎل ﻧﺷﺎن داده ﺧواھد ﺷد‪ ،‬و ﺿرﺑﺎن‬‫ﻗﻠب ﻧﯾز ﻣﻌﻣوﻻً ﺳرﯾﻊﺗر از ﺣﺎﻟت ﻧرﻣﺎل ﻣﯽﺑﺎﺷد‪.‬‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد ﻣرﺗﺑﺎ ً در ﺣﺎل ﺗﻐﯾﯾر اﺳت و ﺑﮫ ﮐﺎری ﮐﮫ اﻧﺟﺎم ﻣﯽدھد و ﻏذاﯾﯽ ﮐﮫ ﺧورده اﺳت‬‫ﺑﺳﺗﮕﯽ دارد‪ .‬آﻧﭼﮫ ﻧوﺷﯾدهاﯾد ﻣﯽﺗواﻧد ﺗﺄﺛﯾر ﺑﺳﯾﺎر ﻗوی و ﺳرﯾﻌﯽ روی ﻓﺷﺎر ﺧوﻧﺗﺎن داﺷﺗﮫ ﺑﺎﺷد‪.‬‬‫ﻣﺑﻧﺎی اﻧدازهﮔﯾریھﺎی اﯾن دﺳﺗﮕﺎه ﺿرﺑﺎن ﻗﻠب ﻣﯽﺑﺎﺷد‪ .‬اﮔر ﺿرﺑﺎن ﻗﻠب ﺑﺳﯾﺎر ﺿﻌﯾف ﯾﺎ ﻧﺎﻣﻧظﻣﯽ‬‫داﺷﺗﮫ ﺑﺎﺷﯾد‪ ،‬ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫﺳﺧﺗﯽ ﺑﺗواﻧد ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﺷﺧﯾص دھد‪.‬‬‫اﮔر دﺳﺗﮕﺎه وﺟود ﺷراﯾط ﻏﯾر طﺑﯾﻌﯽ را ﺗﺷﺧﯾص دھد‪ ،‬اﻧدازهﮔﯾری را ﻣﺗوﻗف ﻣﯽﮐﻧد و ﻧﻣﺎد ﺧطﺎ ظﺎھر‬‫ﻣﯽﺷود‪ .‬ﺑرای ﻣﺷﺎھده ﺗوﺿﯾﺢ ﻧﻣﺎدھﺎ‪ ،‬ﺑﮫ ﺻﻔﺣﮫ ‪ 6‬ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﭘﺎﯾش ﻓﺷﺎر ﺧون ﺑرای اﺳﺗﻔﺎده ﺑزرﮔﺳﺎﻻن طراﺣﯽ ﺷده اﺳت‪ .‬اﮔر ﻣﯽﺧواھﯾد از اﯾن دﺳﺗﮕﺎه‬‫ﺑرای ﮐودﮐﺎن اﺳﺗﻔﺎده ﮐﻧﯾد‪ ،‬ﺑﺎ ﭘزﺷﮑﺗﺎن ﻣﺷورت ﮐﻧﯾد‪ .‬ﮐودﮐﺎن ﻧﺑﺎﯾد ﺑدون ﮐﻣﮏ ﺑزرﮔﺳﺎﻻن از اﯾن‬‫دﺳﺗﮕﺎه اﺳﺗﻔﺎده ﮐﻧﻧد‪.‬‬‫ﻋﻣﻠﮑرد دﺳﺗﮕﺎه ﺧودﮐﺎر ﭘﺎﯾش ﻓﺷﺎر ﺧون ﻣﯽﺗواﻧد ﺗﺣت ﺗﺄﺛﯾر دﻣﺎ ﯾﺎ رطوﺑت ﺷدﯾد ﯾﺎ ارﺗﻔﺎع ﻗرار ﮔﯾرد‪.‬‬‫اﻧدازهﮔﯾری ‪TriCheckTM‬‬‫ﺣﺎﻟت ‪ TriCheckTM‬ﺑﮫ طور ﺧودﮐﺎر و ﭘﯾﺎﭘﯽ ‪ ،‬ﺳﮫ ﺑﺎر ﻋﻣﻠﯾﺎت اﻧدازهﮔﯾری را اﻧﺟﺎم داده و ﻣﯾﺎﻧﮕﯾن‬‫آﻧﮭﺎ را ﻧﻣﺎﯾش ﻣﯽدھد‪.‬‬‫اﻧﺗﺧﺎب ﺣﺎﻟت ‪TriCheckTM‬‬‫‪.1‬‬‫در ﺣﺎﻟت ﺗﻧظﯾم ﻓﺷﺎر‪ ،‬ﺟﮭت ورود ﺑﮫ ﺣﺎﻟت ﺗﻧظﯾم‬‫‪ ،TriCheckTM‬دﮐﻣﮫ ‪ SET‬را ﻓﺷﺎر دھﯾد‪ .‬ﺣﺎﻟت‬‫ﺗﻧظﯾم ﺷده ﻓﻌﻠﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫‪.2‬‬‫ﺑﺎ اﺳﺗﻔﺎده از دﮐﻣﮫھﺎی‬‫ﯾﺎ ﻣﯽﺗواﻧﯾد ﺣﺎﻟت‬‫‪ TriCheckTM‬را ﺧﺎﻣوش ﯾﺎ روﺷن ﮐﻧﯾد‪.‬‬‫روﺷن‪ :‬ﺣﺎﻟت ‪TriCheckTM‬‬‫ﺧﺎﻣوش‪ :‬ﺣﺎﻟت اﻧدازهﮔﯾری ﻋﺎدی )ﺣﺎﻟت ﭘﯾشﻓرض(‬‫‪.3‬‬‫ﺑرای ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه‪ ،‬دﮐﻣﮫ ‪ START‬ﯾﺎ‬‫‪ SET‬را ﻓﺷﺎر دھﯾد‪ .‬ﭘس از ﺳﮫ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از‬‫دﺳﺗﮕﺎه‪ ،‬ﺧودش ﺑﮫطور ﺧودﮐﺎر ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫در ﺣﺎﻟت ﺗﻧظﯾم ﻓﺷﺎر‪،‬‬‫را ﻓﺷﺎر دھﯾد‬‫ﺣﺎﻟت ﺗﻧظﯾم ﺷده ﻓﻌﻠﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‬‫ﯾﺎ‬‫ﺗﻧظﯾم را ﺗﻐﯾﯾر دھﯾد‬‫‪START‬‬‫ﻓﺎرﺳﯽ ‪16‬‬‫ﯾﺎ‬‫اﻧدازهﮔﯾری ﺑﺎ اﺳﺗﻔﺎده از ﺣﺎﻟت ‪TriCheckTM‬‬‫‪.1‬‬‫دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد‪.‬‬‫ھﻣﮫ ﺑﺧشھﺎی ﻧﻣﺎﯾﺷﮕر ﻧﻣﺎﯾﺎن ﻣﯽﺷوﻧد‪.‬‬‫ﺻﻔر )‪ (0‬ﻧﻣﺎﯾﺎن ﻣﯽﺷود و اﻧدﮐﯽ ﭼﺷﻣﮏ ﻣﯽزﻧد‪،‬‬‫ﺳﭘس اﻧدازهﮔﯾری اول آﻏﺎز ﻣﯽﺷود‪.‬‬‫‪.2‬‬‫وﻗﺗﯽ اﻧدازهﮔﯾری ﺗﻣﺎم ﺷد‪ ،‬ﻗراﺋت ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ و‬‫دﯾﺎﺳﺗوﻟﯽ و ﺿرﺑﺎن ﻧﺑض ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‪ ،‬ﺳﭘس‬‫ﺷﻣﺎرش ﻣﻌﮑوس ﯾﮏ دﻗﯾﻘﮫای ﺑرای ﺷروع‬‫اﻧدازهﮔﯾری دوم آﻏﺎز ﻣﯽﺷود‪.‬‬‫‪.3‬‬‫ﭘس از ﮔذﺷت ﯾﮏ دﻗﯾﻘﮫ‪ ،‬اﻧدازهﮔﯾری دوم آﻏﺎز‬‫ﻣﯽﺷود‪.‬‬‫‪.4‬‬‫وﻗﺗﯽ اﻧدازهﮔﯾری ﺗﻣﺎم ﺷد‪ ،‬ﻣﯾﺎﻧﮕﯾن ﻗراﺋتھﺎی ﻓﺷﺎر‬‫اوﻟﯾن و دوﻣﯾن اﻧدازهﮔﯾری ﻧﻣﺎﯾش داده ﻣﯽﺷود‪،‬‬‫ﺳﭘس ﺷﻣﺎرش ﻣﻌﮑوس ﯾﮏ دﻗﯾﻘﮫای ﺑرای ﺷروع‬‫اﻧدازهﮔﯾری ﺳوم آﻏﺎز ﻣﯽﺷود‪.‬‬‫‪.5‬‬‫ﭘس از ﮔذﺷت ﯾﮏ دﻗﯾﻘﮫ‪ ،‬اﻧدازهﮔﯾری ﺳوم آﻏﺎز‬‫ﻣﯽﺷود‪.‬‬‫‪ .6‬وﻗﺗﯽ اﻧدازهﮔﯾری ﺗﻣﺎم ﺷد‪ ،‬ﻣﯾﺎﻧﮕﯾن ﻗراﺋتھﺎی ﺳﮫ‬‫اﻧدازهﮔﯾری ﻧﻣﺎﯾش دادهﺷده و در ﺣﺎﻓظﮫ ذﺧﯾره‬‫ﻣﯽﺷود‪.‬‬‫ﻧﮑﺎت‪:‬‬‫در طول ﻓراﯾﻧد اﻧدازهﮔﯾری‪ ،‬ﻧﻣﺎد ‪TriCheck‬‬‫ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫ﺑرای ﻟﻐو اﻧدازهﮔﯾری‪ ،‬دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر‬‫دھﯾد‪ .‬در اﯾن ﺣﺎﻟت‪ ،‬ھﯾﭻ اطﻼﻋﺎﺗﯽ در ﺣﺎﻓظﮫ‬‫ذﺧﯾره ﻧﻣﯽﺷود‪.‬‬‫در ﺻورﺗﯽ ﮐﮫ ﻧﻣﺎد ﺧطﺎی اﻧدازه ﺑﺎزوﺑﻧد در‬‫اوﻟﯾن اﻧدازهﮔﯾری ظﺎھر ﺷد‪ ،‬اﻧدازهﮔﯾری را ﻟﻐو‬‫ﮐﻧﯾد‪ ،‬ﻣﺟددا ً ﺑﺎزوﺑﻧد را ﺑﮫ طور ﺻﺣﯾﺢ ﺑﺑﻧدﯾد و‬‫اﻧدازهﮔﯾری ﺟدﯾدی را ﺷروع ﮐﻧﯾد‬‫ﭘس از اﺗﻣﺎم اﻧدازهﮔﯾری‪ ،‬ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت‬‫ﻣرﺑوط ﺑﮫ ﺳﮫ اﻧدازهﮔﯾری در ﺣﺎﻓظﮫ ذﺧﯾره‬‫ﻣﯽﺷود‪.‬‬‫در ﺻورت ﻟﻐو اﻧدازهﮔﯾریھﺎ ﻗﺑل از ﺗﮑﻣﯾل‬‫ﺳوﻣﯾن اﻧدازهﮔﯾری‪ ،‬ھﯾﭻ اطﻼﻋﺎﺗﯽ در ﺣﺎﻓظﮫ‬‫ذﺧﯾره ﻧﻣﯽﺷود‪.‬‬‫ھمراﺳﺗﺎ ﺑﺎ ﻗﻠب‬‫را ﻓﺷﺎر دھﯾد‬‫ﮐﻠﯾﮫ ﺑﺧشھﺎی‬‫ﻧﻣﺎﯾﺷﮕر‬‫ﺷ ﻧ‬‫ﻧ ﺎش ا‬‫اوﻟﯾن اﻧدازهﮔﯾری‬‫ﻧﺗﺎﯾﺞ اوﻟﯾن‬‫اﻧدازهﮔﯾری‬‫ﯾﮏ دﻗﯾﻘﮫ ﺑﻌد‬‫دوﻣﯾن اﻧدازهﮔﯾری‬‫‪TM‬‬‫ﻓﺎرﺳﯽ ‪17‬‬‫‪START‬‬‫ﻣﯾﺎﻧﮕﯾن اﻧدازهﮔﯾری‬‫اول و دوم‬‫ﯾﮏ دﻗﯾﻘﮫ ﺑﻌد‬‫ﺳوﻣﯾن اﻧدازهﮔﯾری‬‫ﻣﯾﺎﻧﮕﯾن ﺳﮫ‬‫اﻧدازهﮔﯾری‬‫ﻓراﺧواﻧﯽ اطﻼﻋﺎت ﺣﺎﻓظﮫ‬‫ﻧﮑﺗﮫ‪:‬‬‫اﯾن دﺳﺗﮕﺎه ﻣﯽﺗواﻧد ﺑﮫ طور ﺧودﮐﺎر‪ ،‬ﻧود اﻧدازهﮔﯾری آﺧر را در ﺣﺎﻓظﮫاش ذﺧﯾره ﮐﻧد‪.‬‬‫‪ .1‬دﮐﻣﮫ ﯾﺎ را ﻓﺷﺎر دھﯾد‪.‬‬‫ﻣﯾﺎﻧﮕﯾن ﺗﻣﺎم اﻧدازهﮔﯾریھﺎ و ﺗﻌداد اطﻼﻋﺎت ذﺧﯾره‬‫ﺷده‪ ،‬ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫)در ﺻورﺗﯽ ﮐﮫ ھﯾﭻ اطﻼﻋﺎﺗﯽ ﻣوﺟود ﻧﺑﺎﺷد‪"0" ،‬‬‫ﻧﻣﺎﯾش داده ﻣﯽﺷود‪ .‬ﺑرای ﺧﺎﻣوش ﮐردن دﺳﺗﮕﺎه‪ ،‬دﮐﻣﮫ‬‫‪ ،‬ﯾﺎ ‪ START‬را ﻓﺷﺎر دھﯾد‪(.‬‬‫را ﻓﺷﺎر دھﯾد‬‫ﻣﯾﺎﻧﮕﯾن ﺳﯾﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن دﯾﺎﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن ﺿرﺑﺎن‬‫ﻣﯾﺎﻧﮕﯾن‬‫ﺗﻣﺎم اطﻼﻋﺎت‬‫‪ .2‬ھر ﺑﺎر ﮐﮫ دﮐﻣﮫ )ﯾﺎ دﮐﻣﮫ ﺟﮭت ﻧﻣﺎﯾش‬‫اطﻼﻋﺎت ﺑﮫ ﺗرﺗﯾب ﻣﻌﮑوس( ﻓﺷﺎر داده ﺷود‪،‬‬‫اطﻼﻋﺎت ﺣﺎﻓظﮫ ﺑﮫ ﺷﮑل زﯾر ﻧﻣﺎﯾش داده ﻣﯽﺷوﻧد‪.‬‬‫ﻣﯾﺎﻧﮕﯾن ﺗﻣﺎم اﻧدازهﮔﯾریھﺎی ‪) AM‬ﺻﺑﺢ( ﮐﮫ ﺑﯾن‬‫ﺳﺎﻋﺎت ‪ 4:00‬ﺗﺎ ‪ 9:59‬اﻧﺟﺎم ﺷده اﺳت‪.‬‬‫)در ﻣﺛﺎل ﻧﺷﺎن داده ﺷده‪ 10 ،‬اﻧدازهﮔﯾری‪ .‬در‬‫ﺻورﺗﯽ ﮐﮫ ھﯾﭻ اطﻼﻋﺎﺗﯽ ﻣوﺟود ﻧﺑﺎﺷد‪"--" ،‬‬‫ﻧﻣﺎﯾش داده ﻣﯽﺷود‪(.‬‬‫ﻣﯾﺎﻧﮕﯾن ﺳﯾﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن دﯾﺎﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن ﺿرﺑﺎن‬‫ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت‬‫‪) AM‬ﺻﺑﺢ(‬‫ﻣﯾﺎﻧﮕﯾن ﺗﻣﺎم اﻧدازهﮔﯾریھﺎی ‪) PM‬ﺷب( ﮐﮫ ﺑﯾن‬‫ﺳﺎﻋﺎت ‪ 18:00‬ﺗﺎ ‪ 1:59‬اﻧﺟﺎم ﺷده اﺳت‪.‬‬‫)در ﻣﺛﺎل ﻧﺷﺎن داده ﺷده‪ 9 ،‬اﻧدازهﮔﯾری‪ .‬در‬‫ﺻورﺗﯽ ﮐﮫ ھﯾﭻ اطﻼﻋﺎﺗﯽ ﻣوﺟود ﻧﺑﺎﺷد‪"--" ،‬‬‫ﻧﻣﺎﯾش داده ﻣﯽﺷود‪(.‬‬‫ﺟدﯾدﺗرﯾن اطﻼﻋﺎت )‪ No.n‬در ﻣﺛﺎل ﻧﺷﺎن داده‬‫ﺷده ‪(No.35‬‬‫ﺳﮫ ﺛﺎﻧﯾﮫ ﭘس از ﻧﻣﺎﯾش ﺷﻣﺎره اطﻼﻋﺎت‪ ،‬اطﻼﻋﺎت‬‫اﻧدازهﮔﯾری ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫ﻣﯾﺎﻧﮕﯾن ﺳﯾﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن دﯾﺎﺳﺗوﻟﯽ‬‫ﻣﯾﺎﻧﮕﯾن ﺿرﺑﺎن‬‫ﻣﯾﺎﻧﮕﯾن اطﻼﻋﺎت‬‫‪) PM‬ﺷب(‬‫ﺳﯾﺳﺗوﻟﯽ‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﺿرﺑﺎن‬‫ﺟدﯾدﺗرﯾن داده‬‫ﻗدﯾﻣﯽﺗرﯾن اطﻼﻋﺎت )‪(No.1‬‬‫ﺳﮫ ﺛﺎﻧﯾﮫ ﭘس از ﻧﻣﺎﯾش ﺷﻣﺎره اطﻼﻋﺎت‪ ،‬اطﻼﻋﺎت‬‫اﻧدازهﮔﯾری ﻧﻣﺎﯾش داده ﻣﯽﺷود‪.‬‬‫‪ .3‬ﭘس از اﯾﻧﮑﮫ ﻗدﯾﻣﯽﺗرﯾن اطﻼﻋﺎت ﻧﻣﺎﯾش داده ﺷد‪ ،‬ﺟﮭت‬‫ﺑﺎزﮔﺷت ﺑﮫ ﺻﻔﺣﮫ ﻧﻣﺎﯾش ﻣﯾﺎﻧﮕﯾن ﺗﻣﺎم اﻧدازهﮔﯾریھﺎ‪،‬‬‫دﮐﻣﮫ را ﻓﺷﺎر دھﯾد‪.‬‬‫‪ .4‬دﮐﻣﮫ ‪ START‬را ﻓﺷﺎر دھﯾد ﺗﺎ دﺳﺗﮕﺎه ﺧﺎﻣوش ﺷود‪.‬‬‫ﭘس از ﯾﮏ دﻗﯾﻘﮫ ﻋدم اﺳﺗﻔﺎده از دﺳﺗﮕﺎه‪ ،‬ﺧودش ﺑﮫطور‬‫ﺧودﮐﺎر ﺧﺎﻣوش ﺧواھد ﺷد‪.‬‬‫ﻓﺎرﺳﯽ ‪18‬‬‫ﯾﺎ‬‫ﺳﯾﺳﺗوﻟﯽ‬‫دﯾﺎﺳﺗوﻟﯽ‬‫ﺿرﺑﺎن‬‫آﺧرﯾن داده )ﻗدﯾﻣﯽﺗرﯾن(‬‫ﺷﺎﺧص ‪ IHB/AFib‬ﭼﯾﺳت؟‬‫وﻗﺗﯽ دﺳﺗﮕﺎه رﯾﺗم ﻧﺎﻣﻧظﻣﯽ را در طول اﻧدازهﮔﯾری ﺷﻧﺎﺳﺎﯾﯽ ﮐﻧد‪ ،‬ﻧﺷﺎﻧﮕر ‪ IHB/AFib‬روی ﻧﻣﺎﯾﺷﮕر‬‫ھﻣراه ﺑﺎ ﻣﻘﺎدﯾر اﻧدازهﮔﯾری ظﺎھر ﻣﯽﺷود‪.‬‬‫ﺑﺎ ﭘزﺷﮑﺗﺎن ﺗﻣﺎس‬‫ﻧﮑﺗﮫ‪ :‬ﺗوﺻﯾﮫ ﻣﯽﺷود ﮐﮫ در ﺻورت ﻣﺷﺎھده ﻣﮑرر ﺷﺎﺧص ‪IHB/AFib‬‬‫ﺑﮕﯾرﯾد‪.‬‬‫‪ AFib‬ﭼﯾﺳت؟‬‫ﻗﻠب ﺑﮫ دﻟﯾل ﺳﯾﮕﻧﺎلھﺎی اﻟﮑﺗرﯾﮑﯽ ﮐﮫ در ﻗﻠب روی ﻣﯽدھد‪ ،‬ﻣﻧﻘﺑض ﻣﯽﺷود و ﺧون را ﺑﮫ ﮐل ﺑدن‬‫ﻣﯽﻓرﺳﺗد‪ .‬ﻓﯾﺑرﯾﻼﺳﯾون دھﻠﯾزی )‪ (AFib‬زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ ﺳﯾﮕﻧﺎل اﻟﮑﺗرﯾﮑﯽ در دھﻠﯾز ﻗﻠب‬‫ﻣﻐﺷوش ﺷده و ﻣﻧﺟر ﺑﮫ آﺷﻔﺗﮕﯽ در ﻓواﺻل ﺿرﺑﺎن ﻣﯽﺷود‪ AFib .‬ﻣﯽﺗواﻧد ﻣوﺟب ﺷود ﮐﮫ ﺧون در‬‫ﻗﻠب از ﺣرﮐت ﺑﺎزاﯾﺳﺗد ﮐﮫ ﺑﮫﺳﺎدﮔﯽ ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﻟﺧﺗﮫ ﺧون و ﺑﮫ ﺗﺑﻊ آن ﻣوﺟب ﺳﮑﺗﮫ و ﺣﻣﻠﮫ‬‫ﻗﻠﺑﯽ ﺷود‪.‬‬‫ﻧوار ﻧﺷﺎﻧﮕر ﻓﺷﺎر‬‫اﯾن ﻧﺷﺎﻧﮕر ﻣﯾزان ﻓﺷﺎر در طول اﻧدازهﮔﯾری را ﻧﺷﺎن ﻣﯽدھد‪.‬‬‫در ﺣﺎل اﻓﺰاﯾﺶ ﻓﺸﺎر‬‫آزاد ﺷﺪن ھﻮا‬‫اﻧﺪازهﮔﯿﺮی در ﺣﺎل اﻧﺠﺎم اﺳﺖ‬‫ﺑﺎد ﮐﺮدن در ﺣﺎل اﻧﺠﺎم اﺳﺖ‬‫اﺗﻤﺎم ﺑﺎد ﮐﺮدن‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫ھر ﺑﺧش از ﻧوار ﻧﺷﺎﻧﮕر ﻣﺗﻧﺎظر ﺑﺎ ﯾﮏ ﺑﺧش از طﺑﻘﮫﺑﻧدی‬‫ﻓﺷﺎر ﺧون ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪ (WHO‬اﺳت‪.‬‬‫ﻣﺛﺎل‪:‬‬‫ﺷﺎﺧص طﺑﻘﮫﺑﻧدی ﺳﺎزﻣﺎن ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪(WHO‬‬‫ﻓﺷﺎر ﺧون ﺷدﯾد‬‫ﻓﺷﺎر ﺧون ﻣﺗوﺳط‬‫ﻓﺷﺎر ﺧون ﺧﻔﯾف‬‫ﻧرﻣﺎل رو ﺑﮫ ﺑﺎﻻ‬‫ﻧرﻣﺎل‬‫ﺑﮭﯾﻧﮫ‬‫ﻓﺷﺎر ﺧون ﻣﺗوﺳط‬‫ﻓﺷﺎر ﺧون ﺧﻔﯾف‬‫‪ :‬اﯾن ﺷﺎﺧص ﯾﮏ ﺑﺧش را ﻧﻣﺎﯾش ﻣﯽدھد ﮐﮫ ﺑر ﻣﺑﻧﺎی‬‫اطﻼﻋﺎت ﮐﻧوﻧﯽ و ﻣﺗﻧﺎﺳب ﺑﺎ طﺑﻘﮫﺑﻧدی ‪ WHO‬ﻣﯽﺑﺎﺷد‪.‬‬‫‪:‬‬‫ﻧرﻣﺎل رو ﺑﮫ ﺑﺎﻻ‬‫ﻓﺎرﺳﯽ ‪19‬‬‫در ﻣورد ﻓﺷﺎر ﺧون‬‫ﻓﺷﺎر ﺧون ﭼﯾﺳت؟‬‫ﻓﺷﺎر ﺧون ﻧﯾروﯾﯽ اﺳت ﮐﮫ ﺧون ﺑﮫ دﯾواره رگھﺎ وارد ﻣﯽﮐﻧد‪ .‬ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ‬‫ﻗﻠب ﻣﻧﻘﺑض ﻣﯽﺷود‪ .‬ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ زﻣﺎﻧﯽ روی ﻣﯽدھد ﮐﮫ ﻗﻠب ﻣﻧﺑﺳط ﻣﯽﺷود‪ .‬ﻓﺷﺎر ﺧون ﺑر ﺣﺳب‬‫ﻣﯾﻠﯽﻣﺗر ﺟﯾوه )‪ (mmHg‬اﻧدازهﮔﯾری ﻣﯽﺷود‪ .‬ﻓﺷﺎر ﺧون طﺑﯾﻌﯽ ﯾﮏ ﻓرد ﺗوﺳط ﻓﺷﺎر ﭘﺎﯾﮫ ﻧﺷﺎن داده‬‫ﻣﯽﺷود ﮐﮫ اول ﺻﺑﺢ زﻣﺎﻧﯽ ﮐﮫ ﻓرد ھﻧوز در ﺣﺎل اﺳﺗراﺣت اﺳت و ﻏذاﯾﯽ ﻧﺧورده اﻧدازهﮔﯾری‬‫ﻣﯽﺷود‪.‬‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﭼﯾﺳت و ﭼﮕوﻧﮫ ﮐﻧﺗرل ﻣﯽﺷود؟‬‫ﻓﺷﺎر ﺧون ﺑﺎﻻ ﮐﮫ ﺑﮫ ﻣﻌﻧﺎی ﻓﺷﺎر ﺧون ﻏﯾر ﻧرﻣﺎل ﺑﺎﻻی ﺷرﯾﺎﻧﯽ اﺳت‪ ،‬درﺻورﺗﯽﮐﮫ درﻣﺎن ﻧﺷود‬‫ﻣﯽﺗواﻧد ﻣﺷﮑﻼت زﯾﺎدی را ﺑرای ﻗﻠب از ﺟﻣﻠﮫ ﺳﮑﺗﮫ و ﺣﻣﻠﮫ ﻗﻠﺑﯽ اﯾﺟﺎد ﮐﻧد‪ .‬ﻓﺷﺎر ﺧون ﺑﺎﻻ ﻣﯽﺗواﻧد از‬‫طرﯾﻖ ﺗﻐﯾﯾر ﺳﺑﮏ زﻧدﮔﯽ‪ ،‬ﭘرھﯾز از اﺳﺗرس و ﻣﺻرف دارو ﺗﺣت ﻧظر ﭘزﺷﮏ ﮐﻧﺗرل ﺷود‪.‬‬‫ﺑرای ﭘﯾﺷﮕﯾری از اﺑﺗﻼ ﺑﮫ اﻓزاﯾش ﻓﺷﺎر ﺧون‪ ،‬ﺑﺎﯾد آن را ﺗﺣت ﮐﻧﺗرل داﺷﺗﮫ ﺑﺎﺷﯾد‪:‬‬‫ﻣرﺗب ورزش ﮐﻧﯾد‬‫ﺳﯾﮕﺎر ﻧﮑﺷﯾد‬‫ﺑﮫطور ﻣرﺗب ﺑرای ﭼﮏ آپ و ﻣﻌﺎﯾﻧﮫ ﭘزﺷﮑﯽ‬‫ﻣﺻرف ﻧﻣﮏ و ﭼرﺑﯽ را ﮐﺎھش دھﯾد‬‫ﻣراﺟﻌﮫ ﮐﻧﯾد‬‫وزن ﻣﻧﺎﺳب داﺷﺗﮫ ﺑﺎﺷﯾد‬‫ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ اﻧدازهﮔﯾری ﮐﻧم؟‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون در ﮐﻠﯾﻧﯾﮏ ﯾﺎ ﻣطب ﭘزﺷﮏ ﻣﯽﺗواﻧد ﺑﺎﻋث اﯾﺟﺎد ﺗرس در ﻓرد ﺷده و اﯾن اﻣر‬‫ﻣﯽﺗواﻧد ﺑﺎﻋث اﻓزاﯾش ﻣﻘدار ﻗراﺋت ﺷده ﺗوﺳط دﺳﺗﮕﺎه ﺑﮫ ﻣﯾزان ‪ 25‬ﺗﺎ ‪ mmHg 30‬ﺑﯾش از ﻓﺷﺎر‬‫ﺧون اﻧدازهﮔﯾری ﺷده در ﻣﻧزل ﺷود‪ .‬اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون در ﺧﺎﻧﮫ ﻣوﺟب ﮐﺎھش ﺗﺄﺛﯾر ﻋواﻣل‬‫ﺧﺎرﺟﯽ ﺑر ﻗراﺋت ﻓﺷﺎر ﺧون ﻣﯽﺷود‪ ،‬ﻣﮑﻣل ﻗراﺋت ﭘزﺷﮏ اﺳت و ﺳﺎﺑﻘﮫ دﻗﯾﻖﺗر و ﮐﺎﻣلﺗری از ﻓﺷﺎر‬‫ﺧوﻧﺗﺎن را در اﺧﺗﯾﺎر ﭘزﺷﮏ ﻗرار ﻣﯽدھد‪.‬‬‫طﺑﻘﮫﺑﻧدی ﻓﺷﺎر ﺧون ‪WHO‬‬‫اﺳﺗﺎﻧداردھﺎی ارزﯾﺎﺑﯽ ﻓﺷﺎر ﺧون ﺑﺎﻻ‪،‬‬‫ﺑدون در ﻧظر ﮔرﻓﺗن ﺳن‪ ،‬ﺗوﺳط ﺳﺎزﻣﺎن‬‫ﺑﮭداﺷت ﺟﮭﺎﻧﯽ )‪ (WHO‬طﺑﻖ ﻧﻣودار‪،‬‬‫اراﺋﮫ ﺷدهاﻧد‪.‬‬‫ﻧوﺳﺎﻧﺎت ﻓﺷﺎر ﺧون‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد در طول روز و در‬‫ﻓﺻول ﻣﺧﺗﻠف ﺑﮫ ﺷدت ﺗﻐﯾﯾر ﻣﯽﮐﻧد‪.‬‬‫ﻓﺷﺎر ﺧون ﯾﮏ ﻓرد ﻣﯽﺗواﻧد ﺑر اﺛر‬‫ﺷراﯾط ﻣﺧﺗﻠف در طول روز ﺑﮫ ﻣﯾزان‬‫‪ 30‬ﺗﺎ ‪ mmHg 50‬ﺗﻐﯾﯾر ﮐﻧد‪ .‬در‬‫اﻓرادی ﮐﮫ دﭼﺎر ﻓﺷﺎر ﺧون ﺑﺎﻻ ھﺳﺗﻧد‪ ،‬اﯾن ﻧوﺳﺎﻧﺎت ﺣﺗﯽ ﺑﯾﺷﺗر ﻧﯾز ﻣﯽﺑﺎﺷد‪ .‬ﻣﻌﻣوﻻً ﻓﺷﺎر ﺧون در‬‫ﻣﺣل ﮐﺎر ﯾﺎ در زﻣﺎن ﺑﺎزی اﻓزاﯾش ﯾﺎﻓﺗﮫ و در طول ﺧواب ﺑﮫ ﮐﻣﺗرﯾن ﻣﯾزان ﺧود ﻣﯽرﺳد‪ .‬ﻟذا ﺑﺎ ﻣﺷﺎھده‬‫ﻧﺗﺎﯾﺞ ﯾﮏﺑﺎر اﻧدازهﮔﯾری ﺑﯾش از ﺣد ﻧﮕران ﻧﺷوﯾد‪.‬‬‫ﻓﺎرﺳﯽ ‪20‬‬‫اﻧدازهﮔﯾری ﻓﺷﺎر ﺧون را ھر روز در زﻣﺎن‬‫ﻣﺷﺧﺻﯽ اﻧﺟﺎم داده و از دﺳﺗوراﻟﻌﻣﻠﯽ ﮐﮫ در اﯾن‬‫راھﻧﻣﺎ ﺗوﺿﯾﺢ داده ﺷده اﺳت اﺳﺗﻔﺎده ﮐﻧﯾد ﺗﺎ ﻓﺷﺎر‬‫ﺧون ﻧرﻣﺎﻟﺗﺎن را ﭘﯾدا ﮐﻧﯾد‪ .‬ﻗراﺋت ﻣﻧظم ﻣﯽﺗواﻧد‬‫ﺳﺎﺑﻘﮫ ﺟﺎﻣﻌﯽ از ﻓﺷﺎر ﺧوﻧﺗﺎن را در اﺧﺗﯾﺎرﺗﺎن ﻗرار‬‫دھد‪ .‬ھﻧﮕﺎم ﺛﺑت ﻓﺷﺎر ﺧوﻧﺗﺎن‪ ،‬ﺣﺗﻣﺎ ً ﺗﺎرﯾﺦ و ﺳﺎﻋت‬‫را ﻗﯾد ﮐﻧﯾد‪ .‬ﺑرای ﺗﻔﺳﯾر اطﻼﻋﺎت ﻓﺷﺎر ﺧوﻧﺗﺎن‪ ،‬ﺑﺎ‬‫ﭘزﺷﮏ ﻣﺷورت ﮐﻧﯾد‪.‬‬‫ﻋﯾبﯾﺎﺑﯽ‬‫ﻣﺷﮑل‬‫دﻟﯾل اﺣﺗﻣﺎﻟﯽ‬‫ھﯾﭻ ﭼﯾزی ﻧﻣﺎﯾش داده ﺑﺎﺗریھﺎ ﺧﺎﻟﯽ ﺷدهاﻧد‪.‬‬‫ﻧﻣﯽﺷود‪ ،‬ﺣﺗﯽ زﻣﺎﻧﯽ‬‫ﮐﮫ دﺳﺗﮕﺎه روﺷن‬‫اﻧﺗﮭﺎی ﺑﺎﺗریھﺎ در ﻣوﻗﻌﯾت ﺻﺣﯾﺢ‬‫اﺳت‪.‬‬‫ﻗرار ﻧﮕرﻓﺗﮫاﻧد‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﺎد ﻧﻣﯽﺷود‪.‬‬‫وﻟﺗﺎژ ﺑﺎﺗری ﺧﯾﻠﯽ ﮐم اﺳت‪.‬‬‫)ﻋﻼﻣت ‪LOW‬‬‫‪ (BATTERY‬ﭼﺷﻣﮏ ﻣﯽزﻧد‪.‬‬‫اﮔر ﺑﺎﺗریھﺎ ﺑﮫطور ﮐﺎﻣل ﺧﺎﻟﯽ ﺷده‬‫ﺑﺎﺷﻧد‪ ،‬اﯾن ﻋﻼﻣت ظﺎھر ﻧﻣﯽﺷود‪.‬‬‫ﺑﺎزوﺑﻧد ﺑﮫدرﺳﺗﯽ روی ﺑﺎزو ﯾﺎ ﻣﭻ‬‫ﻗرار داده ﻧﺷده اﺳت‪.‬‬‫در ﺣﯾن اﻧدازهﮔﯾری ﺑﺎزو ﯾﺎ ﺑدﻧﺗﺎن‬‫را ﺗﮑﺎن دادهاﯾد‪.‬‬‫دﺳﺗﮕﺎه اﻧدازهﮔﯾری‬‫ﻧﻣﯽﮐﻧد‪.‬‬‫ﻣﻘﺎدﯾر ﻗراﺋتﺷده ﺧﯾﻠﯽ ﺑﺎزوﺑﻧد در ﻣوﻗﻌﯾت ﺻﺣﯾﺢ ﻗرار‬‫ﺑﺎﻻ ﯾﺎ ﺧﯾﻠﯽ ﭘﺎﯾﯾن‬‫ﻧدارد‪.‬‬‫ھﺳﺗﻧد‪.‬‬‫ﺳﺎﯾر ﻣوارد‬‫ﺗوﺟﮫ‪:‬‬‫ﻋﻣﻠﯾﺎت ﺗوﺻﯾﮫﺷده‬‫ھﻣﮫ ﺑﺎﺗریھﺎ را ﺑﺎ ﺑﺎﺗریھﺎی ﻧو ﺟﺎﯾﮕزﯾن‬‫ﮐﻧﯾد‪.‬‬‫ﺑﺎﺗریھﺎ را ﺧﺎرج ﮐﻧﯾد و آﻧﮭﺎ را ﻣطﺎﺑﻖ ﺑﺎ‬‫ﻋﻼﻣت ﻣﺛﺑت و ﻣﻧﻔﯽ ﮐﮫ در ﻣﺣﻔظﮫ ﺑﺎطری‬‫ﻧوﺷﺗﮫ ﺷده اﺳت‪ ،‬در ﺟﺎی ﺧود ﻗرار دھﯾد‪.‬‬‫ھﻣﮫ ﺑﺎﺗریھﺎ را ﺑﺎ ﺑﺎﺗریھﺎی ﻧو ﺟﺎﯾﮕزﯾن‬‫ﮐﻧﯾد‪.‬‬‫ﺑﺎزوﺑﻧد را ﺑﮫدرﺳﺗﯽ دور ﺑﺎزو ﻗرار دھﯾد‪.‬‬‫ﻣطﻣﺋن ﺷوﯾد ﮐﮫ ﺑدن ﺷﻣﺎ در ﺣﯾن‬‫اﻧدازهﮔﯾری ﺑﯽﺣرﮐت و آرام ﺑﺎﺷد‪.‬‬‫راﺣت و ﺑﯽﺣرﮐت ﺑﻧﺷﯾﻧﯾد‪.‬‬‫دﺳتھﺎﯾﺗﺎن را روی ﻣﯾز ﻗرار دھﯾد ﺑﮫ‬‫ﮔوﻧﮫای ﮐﮫ ﮐف دﺳتھﺎﯾﺗﺎن رو ﺑﮫ ﺑﺎﻻ ﺑوده‬‫و ﺑﺎزوﺑﻧد و ﻗﻠﺑﺗﺎن ھمﺗراز ﺑﺎﺷﻧد‪.‬‬‫اﮔر ﺿرﺑﺎن ﻗﻠب ﺧﯾﻠﯽ ﺿﻌﯾف ﯾﺎ ﻧﺎﻣﻧظﻣﯽ‬‫داﺷﺗﮫ ﺑﺎﺷﯾد‪ ،‬ﻣﻣﮑن اﺳت دﺳﺗﮕﺎه ﺑﮫ ﺳﺧﺗﯽ‬‫ﺑﺗواﻧد ﻓﺷﺎر ﺧوﻧﺗﺎن را ﺗﻌﯾﯾن ﮐﻧد‪.‬‬‫ﻋدد ﻧﺷﺎن داده ﺷده ﺗوﺳط دﺳﺗﮕﺎه‬‫ﺑﮫ ﺑﺧش "ﭼرا ﻓﺷﺎر ﺧون را در ﺧﺎﻧﮫ‬‫ﻣﺗﻔﺎوت از ﻋددی اﺳت ﮐﮫ در‬‫ﮐﻠﯾﻧﯾﮏ ﯾﺎ ﻣطب ﭘزﺷﮏ اﻧدازهﮔﯾری اﻧدازهﮔﯾری ﮐﻧم؟" ﻣراﺟﻌﮫ ﮐﻧﯾد‪.‬‬‫ﺷده اﺳت‪.‬‬‫ﺑﺎﺗریھﺎ را ﺧﺎرج ﮐﻧﯾد‪ .‬آﻧﮭﺎ را ﺑﮫ درﺳﺗﯽ‬‫در ﺟﺎی ﺧود ﻗرار دھﯾد و ﺑﺎر دﯾﮕر‬‫اﻧدازهﮔﯾری را اﻧﺟﺎم دھﯾد‪.‬‬‫اﮔر ﮐﻠﯾﮫ اﻗداﻣﺎت ذﮐرﺷده را اﻧﺟﺎم دادﯾد اﻣﺎ ﻣﺷﮑﻠﺗﺎن ﺣل ﻧﺷد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬ﺳﻌﯽ ﻧﮑﻧﯾد‬‫اﯾن ﻣﺣﺻول را ﺑﺎز ﯾﺎ ﺗﻌﻣﯾر ﮐﻧﯾد‪ ،‬زﯾرا اﯾن ﮐﺎر ﻣوﺟب ﺑﯽاﻋﺗﺑﺎر ﺷدن ﮔﺎراﻧﺗﯽ آن ﻣﯽﺷود‪.‬‬‫ﻓﺎرﺳﯽ ‪21‬‬‫ﻧﮕﮭداری‬‫دﺳﺗﮕﺎه را ﺑﺎز ﻧﮑﻧﯾد‪ .‬اﯾن دﺳﺗﮕﺎه از اﺟزای اﻟﮑﺗرﯾﮑﯽ ظرﯾف و واﺣد ھوای ﭘﯾﭼﯾدهای ﺗﺷﮑﯾل ﺷده اﺳت‬‫ﮐﮫ ﻣﯽﺗواﻧد آﺳﯾب ﺑﺑﯾﻧد‪ .‬اﮔر ﻧﺗواﻧﺳﺗﯾد ﺑﺎ اﺳﺗﻔﺎده از راھﻧﻣﺎی ﻋﯾبﯾﺎﺑﯽ ﻣﺷﮑل دﺳﺗﮕﺎه را ﺣل ﮐﻧﯾد‪ ،‬ﺑﺎ‬‫ﻓروﺷﻧده ﻣﺟﺎز ﺷﮭرﺗﺎن ﯾﺎ ﺑﺧش ﺧدﻣﺎت ﻣﺷﺗرﯾﺎن ﻣﺎ ﺗﻣﺎس ﺑﮕﯾرﯾد‪ .‬اطﻼﻋﺎت ﻓﻧﯽ‪ ،‬ﻗطﻌﺎت و واﺣدھﺎی‬‫ﯾدﮐﯽ از طرﯾﻖ ﺑﺧش ﺧدﻣﺎت ﻣﺷﺗرﯾﺎن ‪ A&D‬در اﺧﺗﯾﺎر ﻓروﺷﻧدﮔﺎن ﻣﺟﺎز ﻗرار ﻣﯽﮔﯾرﻧد‪.‬‬‫اﯾن دﺳﺗﮕﺎه ﺑرای ﺧدﻣﺎتدھﯽ طوﻻﻧﯽ ﻣدت طراﺣﯽ و ﺗوﻟﯾد ﺷده اﺳت‪ .‬اﻣﺎ ﺗوﺻﯾﮫ ﻣﯽﺷود ﮐﮫ ھر ‪ 2‬ﺳﺎل‬‫ﯾﮏﺑﺎر دﺳﺗﮕﺎه را ﺑررﺳﯽ ﮐﻧﯾد ﺗﺎ از ﮐﺎرﮐرد ﺻﺣﯾﺢ و دﻗت آن ﻣطﻣﺋن ﺷوﯾد‪ .‬ﻟطﻔﺎ ً ﺑرای ﺳروﯾس و‬‫ﻧﮕﮭداری دﺳﺗﮕﺎه‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ ﻣﺟﺎز ﯾﺎ ‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫دادهھﺎی ﻓﻧﯽ‬‫ﻧوع‬‫‪UA-1020-W‬‬‫روش اﻧدازهﮔﯾری‬‫اﻧدازهﮔﯾری ﻧوﺳﺎنﺳﻧﺟﯽ‬‫ﻣﺣدوده اﻧدازهﮔﯾری‬‫‪ 0‬ﺗﺎ ‪mmHg 299‬‬‫ﻓﺷﺎر‪:‬‬‫ﻓﺷﺎر ﺳﯾﺳﺗوﻟﯽ‪ 60 :‬ﺗﺎ ‪mmHg 279‬‬‫ﻓﺷﺎر دﯾﺎﺳﺗوﻟﯽ‪ 40 :‬ﺗﺎ ‪mmHg 200‬‬‫‪ 40‬ﺗﺎ ‪ 180‬ﺑﺎر‪ /‬دﻗﯾﻘﮫ‬‫ﺿرﺑﺎن‪:‬‬‫‪mmHg 3±‬‬‫‪%5±‬‬‫دﻗت اﻧدازهﮔﯾری‬‫ﻓﺷﺎر‪:‬‬‫ﺿرﺑﺎن‪:‬‬‫ﻣﻧﺑﻊ ﺗﻐذﯾﮫ‬‫‪ 4‬ﻋدد ﺑﺎﺗری ‪ 1.5‬وﻟﺗﯽ )‪ LR6 ،R6P‬ﯾﺎ ‪ (AA‬ﯾﺎ‬‫آداﭘﺗور )‪) AC (TB-233C‬ﻧدارد(‬‫ﺗﻌداد اﻧدازهﮔﯾریھﺎ‬‫ﺣدود ‪ 1000‬ﻣرﺗﺑﮫ ‪) LR6‬ﺑﺎﺗریھﺎی آﻟﮑﺎﻻﯾن(‬‫ﺣدود ‪ 300‬ﻣرﺗﺑﮫ ‪) R6P‬ﺑﺎﺗریھﺎی ﻣﻧﮕﻧزی(‬‫ﺑﺎ ﻣﻘدار ﻓﺷﺎر ‪ ،mmHg 180‬دﻣﺎی اﺗﺎق ‪ 23‬درﺟﮫ ﺳﺎﻧﺗﯽﮔراد‬‫طﺑﻘﮫﺑﻧدی‬‫ﺗﺟﮭﯾزات ‪ EM‬ﺑﺎ ﺗوان داﺧﻠﯽ )ﺗﻐذﯾﮫ ﺗوﺳط ﺑﺎطری( ‪/‬‬
‫ﮐﻼس ‪) 2‬ﺗﻐذﯾﮫ ﺗوﺳط آداﭘﺗور(‬‫ﺣﺎﻟت ﻋﻣﻠﯾﺎت ﭘﯾوﺳﺗﮫ‬‫آزﻣﺎﯾش ﺑﺎﻟﯾﻧﯽ‬‫ﻣطﺎﺑﻖ اﺳﺗﺎﻧدارد ‪2013：ISO81060-2‬‬‫در ﺑررﺳﯽ اﻋﺗﺑﺎرﺳﻧﺟﯽ ﺑﺎﻟﯾﻧﯽ‪ ،‬از ‪ K5‬ﺟﮭت ﺗﻌﯾﯾن ﻓﺷﺎر ﺧون‬‫دﯾﺎﺳﺗوﻟﯽ ﺑرای ‪ 85‬ﺳوژه اﺳﺗﻔﺎده ﺷد‬‫‪EMD‬‬‫‪IEC 60601-1-2 2014‬‬‫ﺣﺎﻓظﮫ‬‫‪ 90‬اﻧدازهﮔﯾری آﺧر‬‫ﺷراﯾط ﮐﺎر‬‫‪ +10‬ﺗﺎ ‪ / °C +40‬رطوﺑت ﻧﺳﺑﯽ ‪ 15‬ﺗﺎ ‪ 85‬درﺻد ‪ 800 /‬ﺗﺎ‬‫‪ 1060‬ھﮑﺗوﭘﺎﺳﮑﺎل‬‫ﻓﺎرﺳﯽ ‪22‬‬‫ﺷراﯾط ﺣﻣل و ﻧﻘل ‪ /‬ﻧﮕﮭداری‬‫‪ -20‬ﺗﺎ ‪ / °C +60‬رطوﺑت ﻧﺳﺑﯽ ‪ 10‬ﺗﺎ ‪ 95‬درﺻد ‪ 700 /‬ﺗﺎ‬‫‪ 1060‬ھﮑﺗوﭘﺎﺳﮑﺎل‬‫ﺣدود ‪) 140‬ﻋرض( در ‪) 60‬طول( در ‪) 105‬ﻋﻣﻖ( ﻣﯾﻠﯽﻣﺗر‬‫وزن‬‫ﺣدود ‪ 285‬ﮔرم‪ ،‬ﺑدون ﺑﺎﺗریھﺎ‬‫ﻗطﻌﮫ ﺑﮫﮐﺎررﻓﺗﮫ‬‫ﺑﺎزوﺑﻧد ﻧوع ‪BF‬‬‫ﺣﻔﺎظت ورودی‬‫دﺳﺗﮕﺎه‪IP20 :‬‬‫ﻋﻣر ﻣﻔﯾد‬‫دﺳﺗﮕﺎه‪ 5 :‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫ﺑﺎزوﺑﻧد‪ 2 :‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫آداﭘﺗور ‪ 5 :AC‬ﺳﺎل )در ﺷراﯾطﯽ ﮐﮫ ﺷش ﺑﺎر در روز اﺳﺗﻔﺎده ﺷود(‬‫آداﭘﺗور ‪ AC‬ﯾدﮐﯽ‬‫آداﭘﺗور ﺑرای ﻣﺗﺻل ﮐردن دﺳﺗﮕﺎه ﺑﮫ ﻣﻧﺑﻊ ﺑرق ﺧﺎﻧﮕﯽ ﻣﯽﺑﺎﺷد‪.‬‬‫ﻟطﻔﺎ ً ﺑرای ﺧرﯾد‪ ،‬ﺑﺎ ﻓروﺷﻧده ﻣﺣﻠﯽ ‪ A&D‬ﺗﻣﺎس ﺑﮕﯾرﯾد‪.‬‬‫آداﭘﺗور ‪ AC‬ﺑﺎﯾد ﺑﮫﺻورت دورهای ﺑررﺳﯽ و ﺗﻌوﯾض ﺷود‪.‬‬‫ﻧﻣﺎدھﺎﯾﯽ ﮐﮫ روی آداﭘﺗور ‪ AC‬ﭼﺎپ ﺷده اﺳت‬‫ﮐﺎرﮐرد ‪ /‬ﻣﻌﻧﯽ‬‫ﻧﻣﺎدھﺎ‬‫ﺻرﻓﺎ ً ﺑرای اﺳﺗﻔﺎده در ﻣﺣﯾطھﺎی ﺳرﺑﺳﺗﮫ‬‫دﺳﺗﮕﺎه ﮐﻼس دو‬‫ﻓﯾوز ﺣرارﺗﯽ‬‫ﻓﯾوز‬‫ﺑرﭼﺳب دﺳﺗوراﻟﻌﻣل ‪ EC‬دﺳﺗﮕﺎه‬‫ﺑرﭼﺳب ﻣﺟوز ‪ EAC‬دﺳﺗﮕﺎه‬‫ﻗطبھﺎی ﻧری آداﭘﺗور ‪AC‬‬‫اﺑﻌﺎد‬‫‪TB-233C‬‬‫ﻟوازم ﺟﺎﻧﺑﯽ ﮐﮫ ﺟدا ﻓروﺧﺗﮫ ﻣﯽﺷوﻧد‬‫ﺑﺎزوﺑﻧد‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪CUF-F-LA‬‬‫‪CUF-I‬‬‫‪CUF-F-A‬‬‫‪CUF-F-SA‬‬‫آداﭘﺗور ‪AC‬‬‫ﺗوﺟﮫ‪:‬‬‫اﻧدازه ﺑﺎزوﺑﻧد‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﺑزرگ ﺑزرﮔﺳﺎﻻن‬‫ﺑﺎزوﺑﻧد ﻋرﯾض‬‫ﺑﺎزوﺑﻧد ﺑزرﮔﺳﺎﻻن‬‫ﺑﺎزوﺑﻧد ﺳﺎﯾز ﮐوﭼﮏ ﺑزرﮔﺳﺎﻻن‬‫ﺷﻣﺎره ﮐﺎﺗﺎﻟوگ‬‫‪TB-233C‬‬‫اﻧدازه ﺑﺎزو‬‫‪ 31‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 45‬ﺳﺎﻧﺗﯽﻣﺗر‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 42‬ﺳﺎﻧﺗﯽﻣﺗر‬‫‪ 22‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 32‬ﺳﺎﻧﺗﯽﻣﺗر‬‫‪ 16‬ﺳﺎﻧﺗﯽﻣﺗر ﺗﺎ ‪ 24‬ﺳﺎﻧﺗﯽﻣﺗر‬‫دوﺷﺎﺧﮫ‬‫ﻧوع ‪C‬‬‫ﻣﻣﮑن اﺳت ﻣﺷﺧﺻﺎت ﺑدون اطﻼع ﻗﺑﻠﯽ ﺗﻐﯾﯾر داده ﺷوﻧد‪.‬‬‫طﺑﻘﮫﺑﻧدی ‪ IP‬ﻣﯾزان ﺣﻔﺎظت ﻓراھمﺷده ﺗوﺳط ﻣﺣﻔظﮫ ﺗﺟﮭﯾزات ﻣطﺎﺑﻖ ﺑﺎ ‪ IEC 60529‬ﻣﯽﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه در ﺑراﺑر اﺷﯾﺎء ﺧﺎرﺟﯽ ﺟﺎﻣد ﺑﺎ ﻗطر ‪ 12‬ﻣﯾﻠﯽﻣﺗر و ﺑﯾﺷﺗر ﻣﺎﻧﻧد اﻧﮕﺷت ﻣﻘﺎوم ﻣﯽﺑﺎﺷد‪.‬‬‫اﯾن دﺳﺗﮕﺎه در ﺑراﺑر آب ﻣﻘﺎوم ﻧﯾﺳت‪.‬‬‫ﻓﺎرﺳﯽ ‪23‬‬‫ﯾﺎدداﺷت‬‫ﻓﺎرﺳﯽ ‪24‬‬

A&D Company, Ltd.
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119

OX14 1DY

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409
OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司
中国 上海市浦东新区浦东大道138号永华大厦21楼A室 邮编200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599