PH-2B

CT Whole Body Phantom

PBU-60Instruction
manualContents● Please read before training

General information

P. 1 ~ P. 4

● Preparation

Assembly manual

P. 5 ~ P. 10

Please read

General information

Whole body phantom PBU-60
The whole body phantom is life-size human phantom with a life-size synthetic skeleton are embedded
in radiological soft tissue substitute.
There are no metal parts that would impair realistic imaging.The joints, knees, elbows, shoulders andhip-joints, are flexible and able to maintain the necessary positions. The phantom is separable into 10parts facilitating a variety of application. A positioning stand for the head part comes with the set.

Specifications
Whole body phantom PBU-60
Intended application: Positioning phantom for plain radiography
Length: approx.165cm
Weight: approx. 50kg
Can be dismantled at: shoulders elbows, hip joints, knees, neck
Embedded organs: brain, eye balls, lungs with vessels, trachea, mediastinum,
liver with vessels (incl. portal vein), kidneys, spleen, pancreas,
stomach (air), sigmoid colon, rectum, prostate* The product has two 3mm dia holes made in the chest wall to prevent the damage causedby atmospheric pressure change that may occur in transit.

Movable joints and their range of movement
Shoulders: rotate 360 degrees to back and forward, 180 degrees side-ways
Cubital joints (bilateral): bend inward up to 90 degree
Coxae (bilateral): rotate forward up to 90 degrees, then rotate outward up to45 degrees each.
Articulatio genus (bilateral): bend up to 90 degrees.
Materials: Human tissue substitute (urethane, epoxy, etc.,)
Accessories: Positioning head stand 1 piece
Flat head screwdriver 1 piece1 set
Sample images
Hand positioning belt 1 piece
Replacement parts: screws and connection tapes for Patella.

1Please readGeneral information

HU numbers of each organ
HU *numbers

OrgansSoft tissue
Brain

Gall bladder

20

Brain

40

30

Ventricles

10Pancreas
SpleenCerebral artery

280
20

Aorta
Vena cava

40

Kidneys40

Up to the first bifurcation
From the first bifurcation
to the small branches−800
8Heart40

Pulmonary vessels

8Coatal cartilage

90

Liver

HU *numbers

Organs8Eye balls

Trachea

Remarks

Liver
Hepatic veins

70
40

50

Kidneys30

Renal veins

40Ureters

1010Ureters
Urethane foam
Pipe wall’s HU numberShown up to theforth bifurcations

UrinarybladderRemarks

Simulated content

10Bladder wall

30

Semibal vesicle
Prostate

25

Sigmoid colon wall

70

Rectum wall

70

50

Incl. the portal vein

* The Hounsfield numbers shown above are approximate figures of the calculated values and not actual measurement values.

Cerebral artery
Arteries range
3〜4φ
2φ

Middle cerebral artery

Anterior cerebral artery

２1.5φ
1φ0.5φPlease readGeneral information

Please note1) Bone trabeculae are observed only at hands.
2) Contours of the bones are slightly stranger than actual human x-ray image, to facilitate clear
understanding.
3) Basically the phantom is designed to be radiographed under same setting as human body,though, the joints areas may require some adjustment to have better image.

Caution1) Do not make the phantom to take the position unintended by the manufacturer.
It may cause the breakage in the phantom if the joints are forced to bend or rotate to thedirections or degrees of angles outside of designed movable range.
2) Do not remove the plastic protectors on the phantom s hands. The protector prevents
the phantom fingers from breaking.
3) Do not attempt to lift or carry the phantom if the head or limbs are attached to the trunk.
Detach the joints before moving to avoid the excessive force on the joints.
4) When the training with the phantom is not made for a while, pay attention to storage
condition (posture) of the phantom so as to avoid the burden to the connetion area.
For example, condition the head has been resting on the pillow for a while can cause thebreakage to the neck connection area.
5) Do not fasten the screws too tight.
6) The phantom is heavy and could cause damage not only to itself but to people or otherobjects. When you need to handle the assembled phantom, always do it by two or more
people with utmost care.
7) Don’t mark on the phantom with pen or leave printed materials contacted on its surface.
Ink marks on the phantom will be irremovable.

３

Please read

General information

PBU-60 Components
Parts#Parts name

QTYParts#Parts name

QTY

1Head

1

21Connector for left hip joint

1

2

Neck-head connection plate

122

Supporting plate for right hip joint

13Trunk

123

Supporting plate for left hip joint

14

Right upper arm

124

Connector for right knee

15

Left upper arm

125

Connector for left knee

16

Right forearm

126

Screw bolt for neck connection M122

7Left forearm

127

Screw bolt for neck connection
M10×152

8

Right thigh128

Screw bolt for shoulder connection
Screw bolt for parts # 22.23 M 8×1010

9Left thigh129

10Right lower leg

1

11

Left lower leg

1213
14151617181920

430

Screw bolt for elbow joint
M 6×60Screw nut for elbow joint
M6131

Screw bolt for hip joint M8×35

6Right patella

132

Screw bolt for knee joint
M12×20

４

Left patella

133

Connection tape for the patellae

2234Height adjustment panel

1235

Base panel

1

136

Neck connector

1

137

Neck connector fixture

1138

Screw for fixture

1139

Screw bolt M6×182

140

Hand positioning belt

1Connector for shoulder A(Right or left)
Connector for shoulder B(Right or left)
Connector for right elbow(lateral side)
Connector for right elbow(medial side)
Connector for left elbow(lateral side)
Connector for left elbow(medial side)
Connector for right hip joint

４4

Assembly manual

P.5

P.6

P.6

P.8
P.6

P.7

P.6
P.8

P.7

P.9
P.9

The number indicated on the photos above correspond to the page
numbers in this manual.

５Assembly manual※ The circled numbers in the assembly images are parts numbers on the components tableon page 3.

< NECK AND HEAD >1. Remove the two bolts for the neckconnection (part #26) and remove
the neck head connection plate(part #2) from the trunk.

26

226 Screw boltfor neckconnection
M12×20【 Neck sub-assembly 】

2. Screw the neck head connection plate (part #2) on the head (part #1).
127
22727

Screw bolt for neckconnection
M10×15×2【 Connection of the head and the trunk 】

3. Mount the assembled parts (parts #1and #2) to the trunk and secure eachside with a screw bolt (part #26).

６

Assembly manual> ELBOWS >Assemble upper arm parts (part #4 or 5) andforearm parts (part #6 or 7) at joint section.
Fix these parts with elbow connectors(part #16 or 18 and #17 or 19).

30

1917

Left and rightforearm

29 Screw bolt for
elbow joint M 6×60×4
Screw nut for
30 elbow joint M 6×4
29

Connector for left
and right elbow(medial side)4
5

7

Left and rightupper arm

630
18
1629

< SHOULDERS >1. Remove the two clear bolts (part #28)
and remove the external shoulder
connector parts A (part #14) andconnector parts B (part #15).

14

Connector for left
and right elbow(lateral side)28
Screw bolt forshoulderconnection
M 8×10×415
3

2. Then attach the upper arms (parts #4 and #5) at the ball joint to the trunk and place over the twoconnectors for the shoulder (parts #14 and #15). Secure the connectors with the clear bolts(part #28). Follow these steps for both left and right side of the trunk.
28

Left and right upper arm
Caution45

Connector for
14shoulder

15

Screw the bolts after confirming the position of the screw hole.
Be careful not to mix up part #4 and 5. The shape of right and left arms are different.

７Assembly manual< KNEES >1. Connect the thigh (#8, 9) and lower leg (#10, 11) at the knee joint. Insert the dowel of theupper thigh through the hole of knee joint of lower thigh. Set the connector for knee (#24, 25)
to the medial side then screw the knee joint bolt (#32) to fix it.
10 11

89Insert the dowel of the upper thigh throughthe hole of the knee joint of lower leg.
Screw bolt forknee joint

24 25

32

32 Screw bolt forknee joint
M12×20×4

2. After fixing the knee joint, bend the knee and insert the connection tape of the patella (#12, 13)
to the guide aperture of the thigh.
12 13８Assembly manual< HIP JOINTS >20 21

Connector for left
and right hip joint

22 23

1. Remove the three bolts (part #31) toremove the hip joint connectors (parts#20 and #21). The supporting plates(parts #22 and #23) are already attachedat the time of delivery.
31
Screw bolt forhip joint
M8×35×631

2. Mount the thighs (part#8, 9) so that the femur heads fit in the hip sockets and hold themin place with hip joints connectors (part# 20,21).
Each hip joint connector is fixed to the trunk with three screws. Start from the screw in thecenter and move to the sides. Hold the thigh upright when you fix the screw at the rear end.
Be careful not to confuse parts for left and for right.
331
Left and right thigh

8Connector for left
and right hip joint

9９20 21

Assembly manual< HEAD AND STAND >1. Mount the head phantom to the neck connector and fix it with two screws.
136

27 Screw bolt

for neckconnection
M10×15×2

27

2. Assemble the two panels. Attach the height adjustment panel (#34) to thebase panel (#35) at right angle and fix with two screws.
Then, insert the end of the neck connector (#36) in the slit in the panel andfix it with the fixture (#37).
37

39 Screw bolt

M6×18×236

34
35
38

39

39

3. The phantom is ready.
Adjust the height of the neck as needed,by loosen the fixture and slide it throughthe slit. Tighten the fixture again at thedesignated height.

3810Caution■

Don’t mark on the phantom with pen or leave printed materials contacted on its surface.
Ink marks on the phantom will be irremovable.

(World Wide)■

Kyoto Kagaku America Inc.

2020.06

注：モデル表面に取扱説明書や印刷物などが直接触れないようにして下さい。
樹脂表面にインクが転写して消えなくなります。

PH-2B
CT Whole Body Phantom

CT 撮影用全身ファントムPBU-60
取扱説明書

目次

● ご使用の前にご使用前の確認と注意●準備組立について

P. 1 ~ P.４
P. 5 ~ P. 10

ご使用の前にご使用前の確認と注意

PH-2B CT撮影用全身ファントム PBU-60
製品概要
PBU-60は人体近似のX線吸収率をもつ軟組織代替材と人工骨格による等身大の全身ファントムです。
CT値を再現した脳および体幹部の臓器を内蔵しており、人体に近似した全身の画像データが得られるX線CT撮影用のモデルです。四肢の主な関節は人体に近い可動域を持ち、様々な撮影ポジションの設定ができます。また、分解してパーツごとに撮影することもできます。画像の障害になる金属の部品は一切使用しいていません。頭部についてはポジショ二ング用のスタンドが付属しています。

仕様
使用目的：X線単純撮影・X線CT撮影
形式：PBU-60
用途：撮影用人体ファントム
身長：約 165cm
体重：約 50kg
分割できる部位：頸部、両肩、両肘、両股関節部、両膝（10 分割）
可動する関節と可動範囲
両肩：前方向に 360°
横方向に 180°
両肘：90°
両股関節：前方向に 90 °、それぞれ 45°外転
両膝関節：90°
ファントム臓器：脳、脳血管 (左半分)、眼球、肺、気管、縦隔、肝臓、胆嚢、肝血管、脾臓、膵臓、腎臓、胃、S字結腸、直腸、前立腺、精嚢、膀胱、尿管
ファントム骨格：等身大人工骨格※ 運送中の気圧の変化による破損防止のため、予めボディ胸部に 3mm の穴を２箇所設けています。
材 質：軟組織等価材（ポリウレタン樹脂）・骨等価材（エポキシ樹脂）
付属品：頭部固定用台 1 台
交換用ネジ・部品・膝蓋骨用バンド 1 式、マイナスドライバー
サンプル画像 1 式、手のポジション設定用ベルト 1 点11点

ご使用の前にご使用前の確認と注意

臓器関係のCT値
臓器(臓器を構成する組織）各臓器の周り全体の軟組織想定するCT値 ※

40 相当

脳室

10 相当

脳血管

280 相当球20 相当

大 動 脈

40 相当

大 静 脈

40 相当

眼（第一分岐まで） −800 相当

気管支 （第二分岐から第三分岐まで）

心8 相当

臓
管
肺 血

40 相当

肋骨部の軟骨

90 相当

肝 臓

※

考

臓器(臓器を構成する組織） 想定するCT値

8 相当

（全体）
脳備

8 相当

（全体）

70 相当

血管

40 相当

胆嚢

20 相当

膵臓

30 相当

臓

50 相当

（全体）

30 相当

脾腎 臓
尿第一分岐までは発泡ウレタン材を使用し、その先はパイプ状で第三分岐までとする

40 相当
10 相当

管10 相当膀胱内10 相当膀胱壁30 相当

嚢25 相当

前立 腺

50 相当

S字結腸壁70 相当

直腸 壁

70 相当

膀 胱
精第四分岐まで再現

血管
尿管

備 考静脈と門脈を封入※上記の CT 値は目安として設定しているもので撮影の条件等により実測の値とは異なります

模擬脳血管の設定● 血管の太さ
３～４φ
２φ◆ 中大脳動脈◆ 前大脳動脈２1.5φ
１φ0.5φご使用の前にご使用前の確認と注意

ご確認ください
① 指骨、掌骨のみ骨梁を写すことができます。
② このファントムは実際の臨床画像に比べ骨の輪郭がやや強調されるよう意図的に作られています。
③ ファントムには放射線に対し人体に近い特性をもった材質で作られていますが、構造上関節部分の撮影条件は若干生体と異なります。

注意① 破損のおそれがありますので、無理なポジショニングは絶対にしないでください。
② 破損のおそれがありますので、ファントムを移動させるときは必ず分解し、首や関節部分に大きな力がかからないようにしてください。
③ 破損のおそれがありますので、ネジ部品は絶対に強くしめすぎないようにしてください。
④ 手の指は細く破損しやすいため、手につけているプラスチックのカバーは取り外さないでください。
⑤このファントムは、人体と同等の重量があり落下等すると非常に危険です。組み立てて使用する場合や胴部を移動させる場合は、ケガをしないように必ず 2 名以上で取り扱ってください。
⑥ 保管時におきましては、ファントムの接合部位に負荷が掛からない様にして下さい。
特に頭部の下に枕等を置いた状態で長期間保管すると頚部に負担が掛かり破損する恐れがあります。
⑦ サインペン、ボールペン等でモデルの表面に書き込みをしないでください。インクが吸収され消えなくなります。
⑧ モデル表面に印刷物などが直接触れないようにしてください。樹脂にインクが転写され消えなくなります。

３

ご使用の前にご使用前の確認と注意

PBU-60 構成部品
部品
番号

部品名称

個数

部品
番号

部品名称

個数

1

頭部

121左脚接続部品

12

頭･胴部接続部品

122

右脚接続部補強部品

13胴部

123

左脚接続部補強部品

14右上腕

124

右膝接続部品

15

左上腕

125

左膝接続部品

16

右前腕126接続用ボルト

7

左前腕

127

頭･首部接続用ボルト M10×15

28

右大腿

128

肩接続用ボルトM8×10
脚接続部補強部品用(22、23)ボルト

109

左大腿

129

10

右下腿

130

11

左下腿

131

12

右膝蓋骨

132

13

左膝蓋骨

133

膝蓋骨接続用バンド

214

肩接続部品 (左右兼用) A234

上下位置固定柱115

肩接続部品 (左右兼用) B

235

底板

116

右肘接続部品

外側

136

首部接続部品117

右肘接続部品

内側

137

ノブ受け板

118

左肘接続部品

外側

138

高さ固定用ノブ付ネジ

119

左肘接続部品

内側

139

ノブ付ボルト M6×18

2

140

手のポジション設定用ベルト

120

右脚接続部品

４M12x20

肘接続用ボルトM6×6024M64

脚接続用ボルトM8×356

膝接続用ボルトM12×204

肘接続用ナット組立について

P.5

P.6

P.6

P.8
P.6

P.7

P.6
P.8

P.7

P.9
P.9

それぞれの部位の組立は、上の写真で記したページを参照してください。
※各部品の外観は実際の商品とは異なる場合もあります。予めご了承ください。

５

組立について組立手順を紹介していきますが、写真中の部品番号については3ページの構成部品表を参照ください。

< 首部分の部品取り付け >1. ２箇所の頭・胴部接続用ボルト(部品番号26)をはずし、頭･胴部接続部品 (部品番号２)を胴部から取り外します。
2626

2

頭･胴部接続用ボルト
M12×20【 首部分の部品取り付け】
2. 取り外した接続部品を頭部（部品番号１) に組付け、頭･首部接続用ボルト ( 部品番号27 )２本で固定します。
12722727頭･首部接続用ボルト
M10×15×2【 頭部と胴部の組立 】
3. 組立てた頭部 (部品番号１＋２) を胴部の肩のビス２個所に差し込み、頭･胴部接続用ボルト ( 部品番号26 )で固定します。

６

組立について< 肘関節部の組立（写真は左肘関節部）>
1. 上腕部 (部品番号 4,5 ) と前腕部 (部品番号左右肘接続
部品 内側6,7 )を関節部で組み込み、両サイドから外側と内側用の肘接続部品 (部品番号 16,18と17,19 それぞれ各1点)で挟み、肘接続用ボルトとナットで固定します。
29 肘接続用ボルト M 6×60×430 肘接続用ナット M 6×4
2930

19177465

左右前腕左右上腕301816

29

左右肘接続
部品 外側

< 肩部分の組立（写真は左肩部）>
1. 2 箇所のボルトをはずし、胴部から肩接続部品 A ( 部品番号 14) と 肩接続
部品 B ( 部品番号 15) を取り外します。

141528
肩接続用ボルト
M 8×10×4

3 胴部
２. 次に上腕部 ( 部品番号 4,5）を胴部に組み付け、接続部品 A,B で胴体側取付部の円形の突起部を挟み込むようにして組み合わせ、2 本の肩接続用ボルト ( 部品番号 28) で固定します。部品は A (14）、B (15) 共、左右兼用です
2845

1415 左右肩接続部品

上腕部は左右で形状が違いますので、取り違えないようよくお確かめください。
ボルトでの固定は、ネジ穴の位置を確認し、位置を合わせてから行ってください。

７

組立について< 膝部分の組立>（写真は右膝関節部）

1. 大腿部 ( 部品番号 8,9 ) と下腿部 ( 部品番号 10,11) を膝関節部で組み込み、膝部分の内側に膝接続用部品 ( 部品番号 24，25 ) をあてはめ膝接続用ボルト ( 部品番号 32) ２本で 固定します。
10 1189下腿部膝関節部の穴に大腿部関節部のダボを通す

24 253232 膝接続用ボルト
M12×20×4

2. 固定後、膝部を折り曲げ、左右膝蓋骨 ( 部品番号 12,13）に通してある膝蓋骨接続用バンドの先を大腿部下部の穴に差し込むと 膝部接続が完了です。
12 13

８

組立について< 股関節部の組立>（写真は右股関節部）

左右脚接続部品 20 21

左右脚接続部補強部品 22 23

1. ３箇所の脚接続用ボルト(部品番号31)をはずし、脚接続部品 (部品番号 20, 21)を胴部から取り外します。
脚接続部補強部品 (部品番号 22, 23)は分解及びセット時共、胴部から取り外す必要はありません。
3131

脚接続用ボルトM8×35×6

2. 大腿部 (部品番号8, 9) を胴部下部にセットします。大腿上部の球体部分を胴部下部の穴にあてはめ、脚接続部品 (部品番号20,21) を被せ、脚接続用ボルト (部品番号31) で片側３箇所固定します。
3左右大腿部

胴部

8931

左右脚接続部品

20 21

ボルトは中央部から固定し、お尻側のボルトは脚部を前方に持ち上げると、固定しやすくなります。
大腿部は左右で形状が違いますので、取り違えないようよくお確かめください。

９組立について< 頭部固定用台の組立と頭部の接続 >1. 首部接合部品 ( 部品番号 36) と頭部の底部分を、頭・首部接続用ボルト( 部品番号 27) ２本で固定します。
13627 頭･首部接続用ボルト
M10×15×2
27

2. 頭部固定用台を組み立てます。上下位置固定柱 ( 部品番号 34) と底板 ( 部品番号 35) をノブ付ボルト ( 部品番号 39)２本で固定します。
次に首部接合部品 ( 部品番号 36) の端の凸部を上下位置固定柱のスリットに差し込み、ノブ受け板 ( 部品番号 37) を通した高さ固定用ノブ付ネジ ( 部品番号 38) で固定します。

39 ノブ付ボルト
M6×18×2

373436353839

39
3. 頭部のセッティング完了。
※高さ固定用ノブ付ネジ ( 部品番号 38）
で首を持ち上げる高さを調節してください。

3810

モデル表面に取扱説明書や印刷物などが直接触れないようにしてください。
樹脂表面にインクが吸収され消えなくなります。

・ご不明な点は、お買い上げの販売店、もしくは下記 ㈱京都科学まで御連絡ください。

URL : http://www.kyotokagaku.com
e-mail : rw-kyoto@kyotokagaku.co.jp■本社・工場〒612-8388 京都市伏見区北寝小屋町15番地
TEL：075-605 - 2510（直通）FAX：075-605 - 2519

■東京支店

〒113-0033

東京都文京区本郷三丁目26番６号
NREG本郷三丁目ビル2階
TEL：03-3817-8071（直通）FAX：03-3817-8075
2020.06