Special thanks to A&D Australasia Pty Ltd. for creating the original data.

Balances

FG-D Series..................................... Page 36

AD-4212A/B Series.......................... Page 4

FJ Series.......................................... Page 37

AD-4212C Series............................. Page 5

FS-i Series....................................... Page 38

EK Series......................................... Page 6

FS/FS-KL Series.............................. Page 39

EK-G/EW-G Series.......................... Page 7

FV/FW Series................................... Page 40

EK-H Series..................................... Page 8

HC-i Series....................................... Page 41

EK-i/EW-i Series.............................. Page 9

HC/HD Series.................................. Page 42

EP-A Series...................................... Page 10

HJ-150............................................. Page 43

EP-B Series..................................... Page 11

HL Series......................................... Page 44

ER Series......................................... Page 12

HL-i Series....................................... Page 45

EW Series........................................ Page 13

HL-WP Series.................................. Page 46

FA/FX/FY Series.............................. Page 14

HT Series......................................... Page 47

FP Series......................................... Page 15

HV/HW Series.................................. Page 48

FR Series......................................... Page 16

HV-G Series..................................... Page 49

FZ-CT/FX-CT Series........................ Page 17

HVG-WP Series............................... Page 50

FZ-i/FX-i Series................................Page 18

HW-D Series.................................... Page 51

FZ-iWP/FX-iWP Series.................... Page 19

HW-G Series.................................... Page 52

GF Series......................................... Page 20

SF/SG Series................................... Page 53

GF-K Series..................................... Page 21

SJ-H Series...................................... Page 54

GH Series........................................ Page 22

SK Series......................................... Page 55

GP Series......................................... Page 23

SK-WP Series.................................. Page 56

GR Series........................................ Page 24
GX Series......................................... Page 25

Indicators

GX-K Series..................................... Page 26

AD-4316 ........................................... Page 57

HA Series......................................... Page 27

AD-4321 ........................................... Page 58

HF/HF-G/HP Series......................... Page 28

AD-4322........................................... Page 59

HM Series........................................ Page 29

AD-4323 ........................................... Page 60

HR Series......................................... Page 30

AD-4324 ........................................... Page 61

HX Series......................................... Page 31

AD-4325........................................... Page 62
AD-4326/7........................................ Page 63

ScalesAD-4328 ........................................... Page 64

FC Series......................................... Page 32

AD-4329 ........................................... Page 65

FC-i Series....................................... Page 33

AD-4335/35C................................... Page 66

New FG Series................................. Page 34

AD-4401 ........................................... Page 67

FG Series......................................... Page 35

AD-4402 ........................................... Page 682

AD-4404 ........................................... Page 70
AD-4405........................................... Page 71
AD-4406 ........................................... Page 72
AD-4407........................................... Page 73
AD-4408A/C..................................... Page 74
AD-4531/2….................................... Page 75
AD-4532B........................................ Page 76

Others
M Series Moisture Analyzers........... Page 77
AD-4712........................................... Page 78
AD-4713........................................... Page 79
AD-4714........................................... Page 80
AD-4715 ........................................... Page 81
SV/SV-A Series................................ Page 82

3

Simple Calibration Flow Chart
Use breeze break for
0.1 mg & 0.01 mg

Warm up balance forat least 1 hour

Displays CAL 0

Press & hold CAL key
until CALout is displayed

Displays calibration
mass value

Press SAMPLE key

Press SAMPLE key toselect digit, press REZERO key to selectmass or adjust value

Displays CAL 0

Press PRINT key
to enter setting

Displays calibration
mass value

With nothing on weighingpan, press PRINT key

Displays End

Place cal masspress PRINT key

Depends oninternal settings

Balance may output or
store GLP data

End

AD-4212A/B Series
4Simple Calibration Flow Chart
Warm up balance forat least 30 minutes

Displays CAL 0

Press CAL key

Displays calibration
mass value

Press SAMPLE key

Press RE-ZERO key
to select mass value

Displays CAL 0

Press PRINT key
to enter setting

Displays calibration
mass value

With nothing on weighingpan, press PRINT key

Displays End

Place cal masspress PRINT key

Balance returns tonormal weighingmode

Remove cal massEnd

AD-4212C Series
5Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 30 MINUTESDISPLAYS CAL 0

CHECK PAN EMPTY,BALANCE IS IN GRAMSMODE THEN SLIDE CAL
SWITCH ON

STORES ZERO POINTDISPLAYS CAL FPRESS RE-ZERO KEY

DOES SPAN CAL & THENDISPLAYS End

PLACE CAL MASS ON
PAN & PRESS RE-ZERO

BALANCE RETURNS TONORMAL WEIGHINGREMOVE MASS & SLIDECAL SWITCH OFFEND

EK Series
6

Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 10 MINUTESDISPLAY SHOWS 9.801(TYPICAL)PRESS CAL SWITCHDISPLAY SHOWS CAL 0

PRESS ENTER

ZERO CALIBRATION

WHEN STABLE PRESSENTERDISPLAY SHOWS SPANMASS

PLACE MASS ON PAN

DOES SPAN CALWHEN STABLE PRESSENTERDISPLAY SHOWS END

PRESS CALDISPLAY SHOWS ZEROREMOVE MASS & PRESSCALPRESS CAL IF ONLY ZEROCAL REQUIREDEND

EK-G/EW-G Series
7Simple Calibration Flow Chart
REMOVE COVER AND
PRESS CAL SWITCHOR SLIDE SWITCH(IF FITTED)DISPLAYS
009.798 GRAVITYSETTING

PRESS AND HOLD
SAMPLE & PRINT KEYS
UNTIL CALOUTIS DISPLAYEDDISPLAYS CAL 0

ENSURE P AN
EMPTY ANDSTABLE PRESSPRINTDISPLAYSREQUIREDCALIBRATIONMASSPLACE MASS ON
PAN AND PRESSPRINTDISPLAYS ENDREMOVE MASS
AND PRESS CALRETURNS TO
WEIGHING MODE

END

EK-H Series
8Simple Calibration Flow Chart
WARM UP THE
BALANCE FOR
30 MINUTESDISPLAYS
009.798 GRAVITYSETTING

PRESS & HOLDTHE CAL SWITCHDISPLAYS CALRELEASE THE
CAL SWITCHDISPLAYS CAL 0

PRESS PRINT TOPERFORM ZEROCALIBRATIONDISPLAYS A MASSPLACE THE MASSON THE PAN &PRESS PRINTDISPLAYS end
AFTER A FEW SECS.

REMOVE THE MASS
& PRESS CAL OR
MODE

END

EK-i/EW-i Series
9Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 30 MINUTES(BEHIND PLUG R.H.S. OFDISPLAY)DISPLAYS CAL 0

CHECK PAN EMPTY &SLIDE CAL SWITCH ON

STORES ZERO POINTDISPLAYS CAL FPRESS RE-ZERO KEY

DOES SPAN CAL & THENDISPLAYS CAL End

PLACE FULL LOAD MASSON PAN & PRESS REZERO

BALANCE RETURNS TONORMAL WEIGHINGSLIDE CAL SWITCH OFF& REPLACE PLUG

END

EP-A Series
10Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 30 MINUTES(BEHIND PLUG R.H.S. OFDISPLAY)DISPLAYS CAL 0

CHECK PAN EMPTY &SLIDE CAL SWITCH ON

STORES ZERO POINTDISPLAYS CAL FPRESS TARE KEY

DOES SPAN CAL & THENDISPLAYS CAL End

PLACE FULL LOAD MASSON PAN & PRESS TAREBALANCE RETURNS TONORMAL WEIGHINGSLIDE CAL SWITCH OFF& REPLACE PLUG

END

EP-B Series
11Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 30 MINUTESDISPLAYS CAL in

CHECK DISPLAY IS
0.0000 & PRESS CALDISPLAYS CAL ...

WAITDISPLAYS CAL dn

WAIT

LEVER ON SIDE OF
BALANCEDISPLAYS CAL. . .

GENTLY LOWER CALMASSDISPLAYS CAL upWAIT

DISPLAYS CAL ...

GENTLY RAISE CAL MASSDISPLAYS CAL EndWAITDISPLAYS 0.0000

RETURNS TO NORMAL
WEIGHING

END

ER Series
12

Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 30 MINUTESSWITCH THE BALANCEOFF

PLACE A MASS > 50%
CAPACITY ON PANMASS DISPLAYEDSWITCH THE BALANCEONDISPLAY SHOWS ZEROPOINTREMOVE THE MASSDIP SWITCH &T R IMME R

ADJUST FOR BEST ZEROPOINT

DISPLAYS PRECISEZERO

PRESS ZERODISPLAYS SPANREADING

PLACE FULL CAPACITYMASS ON PAN

T R IMME R

ADJUST SPAN

MAY NEED TO LOOP AS
SPAN & ZERO INTERR E ACTREMOVE MASS CHECK
ZERO

END

EW Series
13

Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST 30 MINUTES(AT REAR OF
INSTRUMENT)DISPLAYS CAL 0

CHECK PAN EMPTY &SLIDE CAL SWITCH ON

STORES ZERO POINTDISPLAYS CAL FPRESS RE-ZEROKEY

DOES SPAN CAL & THENDISPLAYS CAL End

PLACE CAL MASS ON
PAN & PRESS RE-ZERO

BALANCE RETURNS TONORMAL WEIGHINGREMOVE MASS & SLIDECAL SWITCH OFFEND

FA/FX/FY Series
14Simple Calibration Flow Chart
ENSURE BALANCE IS
LEVEL & WARMED UP
FOR 1 HOUR

DISPLAYS CAL 0 FOR A
FEW SECONDSENSURE P AN IS EMPTY &THEN PRESS CAL SWITCH

WAIT FOR CAL MASSDISPLAYDISPLAYS CAL F FOR A
FEW SECONDS

PLACE CAL MASS ON
PAN

WAIT FOR CAL EndDISPLAYDISPLAYS BLANKS THENSHOWS 0.0

REMOVE CAL MASSEND

FP Series
15

Simple Calibration Flow Chart
ENSURE BALANCE IS
LEVEL & WARMED UP
FOR 1 HOUR

PRINT MUST BE PRESSEDIMMEDIAYELY AFTERPRESSING CALENSURE P AN IS EMPTY &THEN PRESS CAL SWITCH& THEN PRESS PRINTDISPLAYS CALFOLLOWED BY LAST SET
CAL MASSUSE SAMPLE (+50g),
MODE(+0.0010g) &PRINT (+0.0001g) KEYS

CHANGE CAL MASS IF
REQUIREDSTORES ZERO POINT &THEN DISPLAYS CAL FCHECK PAN EMPTY THEN
PRESS RE-ZERO

STORES SPAN POINT &THEN DISPLAYS CAL End

PLACE CHOSEN CALMASS ON PAN

RETURNS TO NORMAL
WEIGHINGREMOVE CAL MASS

END

FR Series
16

Simple Calibration Flow Chart
Warm up balance forat least 30 minutes

Displays CAL 0

Press and hold CAL key
until CALout is displayed

Full calibration
capacity displayed

Press SAMPLE key
Press SAMPLE key toselect digit, Press ZERO
key to increase digit,
MODE key to decrease digit

Displays CAL 0

Press PRINT toenter setting

Full calibration load
displayed.

With nothing on theweigh pan,Press PRINT“END” displayed

Place test mass
onto weigh pan.
Press PRINT

Balance returns tonormal weighingmode.

Remove test mass.
END

FZ-CT/FX-CT Series
17Simple Calibration Flow Chart
Warm up balance forat least 30 minutes

Displays CAL 0

Press and hold CAL key
until CALout is displayed

Full calibration
capacity displayed

Press SAMPLE key
Press SAMPLE key toselect digit, Press ZERO
key to increase digit,
MODE key to decrease digit

Displays CAL 0

Press PRINT toenter setting

Full calibration load
displayed.

With nothing on theweigh pan,Press PRINT“END” displayed

Place test mass
onto weigh pan.
Press PRINT

Balance returns tonormal weighingmode.

Remove test mass.
END

FZ-i/FX-i Series
18Simple Calibration Flow Chart
Warm up balance forat least 30 minutes

Displays CAL 0

Press and hold CAL key
until CALout is displayed

Full calibration
capacity displayed

Press SAMPLE key
Press SAMPLE key toselect digit, Press ZERO
key to increase digit,
MODE key to decrease digit

Displays CAL 0

Press PRINT toenter setting

Full calibration load
displayed.

With nothing on theweigh pan,Press PRINT“END” displayed

Place test mass
onto weigh pan.
Press PRINT

Balance returns tonormal weighingmode.

Remove test mass.
END

FZ-iWP/FX-iWP Series
19Simple Calibration Flow Chart
PRESS AND HOLDCAL UNTIL CAL out
IS DISPLAYEDRELEASE THE
CAL KEYDISPLAYS CAL 0

ENSURE P AN
EMPTY ANDSTABLE PRESSPRINTDISPLAYSREQUIREDCALIBRATIONMASSPLACE MASS ON
PAN AND PRESSPRINTDISPLAYS End

RETURNS TO
WEIGHING MODE
NOTE IF GLP MODE
IN USE THENRESULTS AREOUTPUT FIRSTREMOVE MASS

END

GF Series
20

Simple Calibration Flow Chart
ENSURE PAN IS
EMPTY

WARM UP BAL ANCEFOR ATLEAST 30 MINUTESUNTIL CALout
IS DISPL AYED
THEN RELEASEPRESS & HOLDCALZERO ST OREDDISPLAYS CALMASSPRESS PRINTDISPLAYS End

PLACE CAL MASS
& PRESS PRINTREMOVE CALMASSDEPENDS ON SETGLP OUTPARAMETERCALIBRATIONREPORT OUTPUTOR STOREDEND

GF-K Series
21Simple Calibration Flow Chart
WARM UPBAL ANCE FOR AT
LEAST 1 HOUR

DISPLAYS Cal out

PRESS & HOLDCAL B UTT ONDISPLAYS Cal 0

RELEASE CAL
BUTT ONZERO POINT
STORED CALMASS DISPLAYEDENSURE P AN
EMPTY & PRESSPRINTCALIBRATIONCOMPLETEDDISPLAYS end

PLACE MASS &PRESS PRINT

DEPENDS ON
INTERNALSETTINGSBAL ANCE MA Y
OUTPUT OR
STORE GLP DATA

END

GH Series
22

Simple Calibration Flow Chart
WARM UP THE
BALANCE FOR
30 MINUTESDISPLAYS
009.798 GRAVITYSETTING

PRESS & HOLDTHE CAL SWITCHDISPLAYS CALout

RELEASE THE
CAL SWITCHDISPLAYS CAL 0

PRESS PRINT TOPERFORM ZEROCALIBRATIONDISPLAYS A MASSPLACE THE MASSON THE PAN &PRESS PRINTDISPLAYS
AFTER A FEW SECS.

REMOVE THE MASS
THE BALANCERETURNS TO
WEIGHING MODE

END

GP Series
23

Simple Calibration Flow Chart
PRESS AND HOLDCAL UNTIL CAL out
IS DISPLAYEDRELEASE THE
CAL KEYDISPLAYS CAL 0

ENSURE P AN
EMPTY ANDSTABLE PRESSPRINTDISPLAYSREQUIREDCALIBRATIONMASSPLACE MASS ON
PAN AND PRESSPRINTDISPLAYS End

RETURNS TO
WEIGHING MODE
NOTE IF GLP MODE
IN USE THENRESULTS AREOUTPUT FIRSTREMOVE MASS

END

GR Series
24

Simple Calibration Flow Chart
PRESS AND HOLDCAL UNTIL CAL out
IS DISPLAYEDRELEASE THE
CAL KEYDISPLAYS CAL 0

ENSURE P AN
EMPTY ANDSTABLE PRESSPRINTDISPLAYSREQUIREDCALIBRATIONMASSPLACE MASS ON
PAN AND PRESSPRINTDISPLAYS End

RETURNS TO
WEIGHING MODE
NOTE IF GLP MODE
IN USE THENRESULTS AREOUTPUT FIRSTREMOVE MASSEND

GX Series
25

Simple Calibration Flow Chart
ENSURE P AN IS
EMPTY

WARM UP BALANCEFOR ATLEAST 30 MINUTESUNTIL CALout
IS DISPLAYED
THEN RELEASEPRESS & HOLDCALZERO STOREDDISPLAYS CALMASSPRESS PRINTDISPL AYS End

PLACE CAL MASS
& PRESS PRINTREMOVE CALMASSDEPENDS ON SETGLP OUTPARAMETERCALIBRATIONREPOR T OUTPUTOR STOREDEND

GX-K Series
26Simple Calibration Flow Chart

WARM UP BALANCE FOR
AT LEAST ONE HOUR &ENSURE P AN IS EMPTYDISPLAY SHOWS CAL
THEN SHOWS LAST CALMASS USEDUSE RANGE, MODE &PRINT TO CHANGE MASSIF REQUIREDPRESS & HOLD CAL KEY
THEN PRESS PRINT KEY

ZERO CALIBRATIONPRESS RE-ZERODISPLAYS CAL 0

DISPLAYS CAL FREADY FOR SPAN
CALIBRATIONPLACE CAL MASS ON
PA N

SPAN CALIBRATIONDISPLAYS CAL FCALIBRATIONCOMPLE TE DDISPLAYS CAL End

RETURNS TO NORMAL
WEIGHINGREMOVE CAL MASSEND

HA Series
27

Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST ONE HOUR

DISPLAY SHOWS CAL 0

PRESS & HOLD RE-ZERONO
SELECT CAL
MASS?

YES
STANDARD CAL MASSDISPLAYEDPRESS MODE

MODE TO SELECT
RE-ZERO TO ADJUSTUSE MODE & RE-ZERO TO
SET MASSSTORES SELECTIONPRESS PRINTENTERS ZERO CALPRESS RE-ZERO

PLACE MASS ON PANENTERS SPAN
CALIBRATION THENDISPLAYS CAL END

PRESS RE-ZEROEND

HF/HF-G/HP Series
28Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST ONE HOUR &ENSURE P AN IS EMPTYDISPLAY SHOWS CAL 0

PRESS & HOLD RE-ZERO

CALIBRATION WITHCHECKRELEASE KEY WHENDISPLAY SHOWS CAL tSt

ENTERING CAL MODEDISPLAYS CC in
PRESS PRINTIMMEDIATELYREADY FOR ZERO CALPRESS MODE, THEN USE
MODE & RE-ZERO TO
CHANGE SPAN CAL MASSIF REQUIRED. PRESSPRINT TO CONTINUEDISPLAYS CC 0

DOES ZERO CALPRESS RE-ZERO

SHOWS MEASUREDDATADISPLAYS-00001g

READY FOR SPAN CALDISPLAYS CC 200

PLACE CAL MASS ON
PAN & PRESS RE-ZERO

SHOWS MEASUREDDATADISPLAYS 2000001g

CALIBRATIONCOMPLE TE DDISPLAYS tSt End

END

HM Series
29

Simple Calibration Flow Chart
WARM UP BALANCE FOR
AT LEAST ONE HOUR &ENSURE P AN IS EMPTYDISPLAY SHOWS CAL 0

PRESS & HOLD REZERO

ENTERING CAL MODEDISPLAYS CAL 0

READY FOR ZERO CALPRESS MODE, THEN USE
MODE & RE-ZERO TO
CHANGE SPAN CAL MASSIF REQUIRED. PRESSPRINT TO CONTINUEDISPLAYS CAL 0

DOES ZERO CALPRESS RE-ZEROREADY FOR SPAN CALDISPLAYS CAL 200

PLACE CAL MASS ON
PAN & PRESS RE-ZERO

DOES SPAN CALDISPLAYS CAL 200

CALIBRATIONCOMPLE TE DDISPLAYS CAL End

RETURNS TO NORMAL
WEIGHINGREMOVE CAL MASS

END

HR Series
30

Simple Calibration Flow Chart

WARM UP BALANCE FOR
AT LEAST ONE HOUR &ENSURE P AN IS EMPTYDISPLAY SHOWS CAL
THEN SHOWS LAST CALMASS USEDUSE MODE & SAMPLE TO
CHANGE MASS IF
REQUIRED

PRESS CAL KEY & THEN
IMMEDIATELY PRESSPRINT KEY

ZERO CALIBRATIONPRESS RE-ZERODISPLAYS CAL FREADY FOR SPAN
CALIBRATIONPLACE CAL MASS ON
PA N

SPAN CALIBRATIONDISPLAYS CAL FCALIBRATIONCOMPLE TE DDISPLAYS CAL End

RETURNS TO NORMAL
WEIGHINGREMOVE CAL MASSEND

HX Series
31

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAY SHOWS CALSLIDE CAL SWITCH ON

DISPLAY SHOWS CAL 0 &SPAN MASSUSE NUMERIC KEYS TOCHANGE IF REQUIREDPRESS ZEROSTORES CAL MASSVALUEPRESS ENTERSTORES ZERO POINT &DISPLAYS CAL F & MASSENSURE PAN EMPTY &PRESS ZERO WHEN
STABLESTORES SPAN POINT

PLACE CAL MASS ON
PAN & PRESS ENTER
WHEN STABLECALIBRATIONCOMPLE TE DSLIDE CAL SWITCH OFF

END

FC Series
32

Simple Calibration Flow Chart
WARM UP SCALE
FOR AT LEAST 30
MINUTESREMOVE CAL SWITCH COVER TOACCESS THE SWITCH.
DISPLAY CALPRESS CAL
SWIT CHDISPLAYS CAL & FLASHESDESIRED CALIBRATION MASSPRESS ZERO KEYDISPL AYS CAL 0

PRESS ENTERDISPL AYS CAL FENSURE P AN
EMPTY & PRESSENTERDISPLAYS
Calibration End

PLACE MASS &PRESS ENTERPRESS CAL

END

FC-i Series
33Simple Calibration Flow Chart
WARM UP SCALE
FOR AT LEAST 30
MINUTESUNTIL DISPL AY SHOWS CAL 0

PRESS & HOLDCAL SWITCHENSURE EMPTYPAN AND WAIT
FOR STABLESHOWS SPn 1 FOLLOWED BY CALMASS (MAX CAPACITY)
USE RE-ZERO (ZERO) & PRINT TO
CHANGE CAL MASSPRESS MODE

PLACE CAL MASSON PAN & WAIT
FOR STABLECALIBRATES &DISPLAYS End

PRESS MODE

END

NEW FG Series
34Simple Calibration Flow Chart
DISABLE AUTO OFFFUNCTION BY PLACINGMASS ON PANWARM UP SCALE FOR AT
LEAST 10 MINUTESDISPLAY SHOWSGRAVITY VALUE( TYPICAL 09.978)

PRESS CAL SWITCHDISPLAY SHOWS CAL 0

PRESS ZEROSTORES ZERO POINTDISPLAY & S CAL 1

ENSURE PAN EMPTY &PRESS ZERO WHEN
STABLEDISPLAYS CAL MASS
USE HIDDEN KEYS TOCHANGE IF REQUIREDPRESS ZEROSTORES SPAN POINT &DISPLAYS End

PLACE CAL MASS ON
PAN & PRESS ZERO
WHEN STABLECALIBRATIONCOMPLE TE DPRESS CAL SWITCH

END

FG Series
35

Simple Calibration Flow ChartNOTES FOR USA.
1. USE TAREKEY NOT ZERO.
2. PRESS ZEROHERETO SET POUNDSCAL MASS SHOWS CAL 2

WARM UP THE
SCALE FOR AT
LEAST 30 MINS.

DISABLE THE
AUTO POWER
OFF

PRESS THECAL SWITCHDISPLAYS THE
GRAVITYSETTING

PRESS THECAL SWITCHDISPLAYSCAL 0

ENSURE P AN
EMPTY & STABLE
PRESS ZERODISPLAYS
CAL 1

PRESS ZERODISPLAYSCAL MASS

PLACE MASS &PRESS ZERO
WHEN STABLEDISPLAYSEnd

PRESS CAL

RETURNS TO
WEIGH MODE

END

FG-D Series
36Simple Calibration Flow ChartDISPLAY SHOWSGRAVITY VALUE(TYPICAL 09.978)

PRESS CAL SWITCHDISPLAY SHOWS CAL 0

PRESS CAL SWITCHDISPLAYS CAL 0 WITH 0
FLASHING
THEN DISPLAYSSUGGESTED CAL MASSENSURE NO LOAD &PRESS ZERO WHEN
STABLEDISPLAYS CAL MASSPRESS AUXILIAR Y
BUTTON TO SELECT CALMASSDISPLAYS CAL FPRESS CAL BUTTONDISPLAYS CAL F WITH F
FLASHING THEN SHOWSEndATTACH CAL MASS &PRESS ZERORESUMES NORMAL
WEIGHING

PRESS ON/OFFEND

FJ Series
37

Simple Calibration Flow Chart
Press & holdcalibration switch.

“CAL” displayede.g. 9.7985 displayed

Press ENT key, thenthe gravity
acceleration value isdisplayed. Adjust if
necessary.

“CAL 0” displayed

Press ENT key.

e.g. º 30.00kg

Once stable with noload, press ENT. The
display shows SPn 1,& then the span
figure.

To calibrate with adifferent value,change the displayed
value using the 10key pad.

Place the test masson the pan and wait
for the stableindicator. Press
ENT, the scalecalibrates the span.

“END” displayed

Remove test massand press the
ON/OFF key twice
to return to normal
weighing mode.

FS-i Series
38Simple Calibration Flow Chart
ENSURE SCALE IS
WARMED UP FOR 30
MINUTESDISPLAYS C kg

ENSURE PAN IS EMPTY &PRESS CAL SWITCHSCALE CAN BE
CALIBRATED WITH KG
OR LB MASSPRESS UNITS TO SELECTCAL UNITDISPLAYS C 0kg

PRESS TAREDISPLAYS ---FOLLOWEDBY C 06kg

ENSURE PAN EMPTY &PRESS POWER ON/OFF

USE NUMERIC KEYS TOSET REQUIRED CAL MASSPLACE CAL MASS ON
PAN & PRESS POWER
ON/OFFDISPLAYS ---FOLLOWEDBY C End kg

PRESS POWER ON/OFF
TO RETURN TO
WEIGHING

END

FS/FS-KL Series
39Simple Calibration Flow Chart
ALLOW 10 MINUTE WARM
UP PERIODPRESS CAL SWITCHDISPLAY SHOWS GRAVITYVALUE

PRESS MODEDISPLAY SHOWS CAL 0

WHEN STABLE PRESS ZEROTHEN PRESS CAL IF
ZERO CAL ONLYREQUIREDYESDISPLAY SHOWS CAL 1(FULL SCALE CAL)

FULL CAPACITYCAL?

NODISPLAY SHOWS CAL 2(2/3 FULL SCALE CAL)PRESS MODE TO SELECTCAL 2PLACE MASS ON PAN

WHEN STABLE PRESS ZEROPRESS CALDISPLAY SHOWS END

E ND

FV/FW Series
40Simple Calibration Flow Chart“CAL” displayed

Press calibration(CAL) switch

Display flashes therequired calibration
mass

Press the ZERO key
to enter the zero &span calibration
mode.

The span value canbe changed via the0

9 10-keypad.

Display e.g.
Count

Press the ENTER
key. The calibration
value stops flashing.

. CAL .
WeightUnit Weighto

5.000kg

CAL 0

With nothing on theweigh pan, press
ENTER key.

“CAL F” displayed

Place test mass ontoweigh pan & press
ENTER key.
The display returns
to step 1.
Remove test mass.

“CAL” displayed

Press ON/OFF toexit CAL mode.

HC-i Series
41Simple Calibration Flow Chart
DISABLE AUTO POWER
OFF BY PLACING MASSON PANWARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAY SHOWS CALPRESS CAL SWITCHDISPLAY SHOWS CAL XX
WHERE XX IS SCALECAPACITY

ENSURE EMPTY PAN &PRESS ZEROUSE SAMPLE KEY TO
CHANGE CAL MASSSELECT CAL MASS TO BEUSEDDISPLAYS CAL 0

ENSURE PAN EMPTY &PRESS ENTERDISPLAYS CAL FDISPLAYS End AFTERCAL COMPLETEDPLACE CAL MASS ON
PAN & PRESS ENTERLEAVES CALIBRATION
MO D EREMOVE MASS & PRESSCAL SWITCHEND

HC/HD Series
42Simple Calibration Flow Chart
ENSURE PAN
EMPTYDISPLAYS CALPRESS & HOLD
LIGHT/CAL KEY
FOR 3 SECSCALIBRATESZERO & DISPLAYS
100.0 gRELEASE KEY

AFTER APPROX 8 SECSDISPLAYS F FOR SUCCESSFULCAL OR FOR FAILURE THEN
AUTOMATICALLY TURNS OFF

PLACE 100G ON
PAN CENTRE
WITHIN 10 SECS

END

HJ-15043

Simple Calibration Flow Chart
DISABLE THE
AUTO POWER
OFF FUNCTION

WARM UP SCALE
FOR AT LEAST 10MINUTESTURN OFF USING
ON/OFF KEY

HIDDEN KEY IS BETWEEN RE-ZERO & ON/OFF

ENSURE PAN EMPTYPRESS & HOLD RE ZERO & HIDDEN KEYDISPL AYS CALPRESS ON/OFFDISPLAYS CAL 0

PRESS RE-ZERO

CALIBRATES ZERODISPLAYS CAL FWAIT FORSTABLEPRESS RE-ZERO

PLACE CAL MASSWAIT STABLE
PRESS RE-ZERODISPL AYS END

END

HL Series
44

Simple Calibration Flow Chart
ENSURE THE
SCALE IS OFF

PRESS & HOLD
RE-ZERO ANDUNITS KEYSCALIS DISPLAYEDPRESS THE
ON/OFF KEY

CAL 0IS DISPLAYEDPRESS THE
RE-ZERO KEYCAL FIS DISPLAYEDENSURE P AN
EMPTY & ST ABLE
PRESS RE-ZEROEnd
IS DISPLAYEDPLACE MASSWAIT STABLE
PRESS RE-ZERO

END

HL-i Series
45Simple Calibration Flow Chart
DISABLE THE
AUTO POWER
OFF FUNCTION

WARM UP SCALE
FOR AT LEAST 10MINUTESTURN OFF USING
ON/OFF KEY

HIDDEN KEY IS SLIGHTLY UP AND TO THE
RIGHT OF THE RE-ZERO KEY

ENSURE PAN EMPTYPRESS & HOLD RE ZERO & HIDDEN KEYDISPL AYS CALPRESS ON/OFFDISPL AYS CAL 0

PRESS RE-ZERO

CALIBRATES ZERODISPL AYS CAL FWAIT FORSTABLEPRESS RE-ZERO

PLACE CAL MASSWAIT STABLE
PRESS RE-ZERODISPL AYS END

END

HL-WP Series
46Simple Calibration Flow Chart
Turn scale ON

Displays CAL

Remove cover &press cal switch

Displays CAL 0

Press RE-ZERO

Performs zero
calibration thendisplays CAL F

Ensure pan empty
wait for STABLE &press RE-ZERO

Displays End

Place cal mass
wait for STABLE &press RE-ZEROEnd

HT Series
47

Simple Calibration Flow Chart
ALLOW 10 MINUTE WARM
UP PERIODPRESS CAL SWITCHDISPLAY SHOWS GRAVITYVALUE

PRESS MODEDISPLAY SHOWS CAL 0

WHEN STABLE PRESS ZEROTHEN PRESS CAL IF
ZERO CAL ONLYREQUIREDYESDISPLAY SHOWS CAL 1(FULL SCALE CAL)

FULL CAPACITYCAL?

NODISPLAY SHOWS CAL 2(2/3 FULL SCALE CAL)PRESS MODE TO SELECTCAL 2PLACE MASS ON PAN

WHEN STABLE PRESS ZERODISPLAY SHOWS END

PRESS CAL TO END

HV/HW Series
48Simple Calibration Flow Chart
PRESS & HOLD THE
CAL SWITCH

CAL 0 IS DISPLAYEDENSURE PAN EMPTYPRESS ENTER WHEN
STABLEDISPLAYS SPN1 THENSCALE MAXIMUMUSE SAMPLE & SET
TO KEY IN INTENDED
CALIBRATION MASSPLACE MASS AND
PRESS ENTER
WHEN STABLEDISPLAYS ENDREMOVE MASS AND
PRESS CAL TOCOMPLETE

HV-G Series
49Simple Calibration Flow Chart
PRESS & HOLD THE
CAL SWITCH

CAL 0 IS DISPLAYEDENSURE PAN EMPTYPRESS ENTER WHEN
STABLEDISPLAYS SPN1 THENSCALE MAXIMUMUSE SAMPLE & SET
TO KEY IN INTENDED
CALIBRATION MASSPLACE MASS AND
PRESS ENTER
WHEN STABLEDISPLAYS ENDREMOVE MASS AND
PRESS CAL TOCOMPLETE

HVG-WP Series
50Simple Calibration Flow Chart

NOTE FOR USA
PRESS MODEHERETO SELECTCALIBRATION IN
POUNDS, CAL 3

WARM UP THE
SCALE FOR AT
LEAST 10 MINS.

DISABLE THE
AUTO POWER
OFF FEATUREREMOVE THE
COVER & PRESSCALDISPLAYS THE
GRAVITYSETTING

PRESS ZERODISPLAYSCAL 0

ENSURE SCALEUNLOADED &STABLE THEN
PRESS ZERO

PLACE CAL MASSPRESS ZERO
WHEN STABLE

PRESS CALDISPLAYSCAL 1

DISPLAYSEnd

RETURNSTO WEIGHING

END

HW-D
51

Simple Calibration Flow Chart
PRESS & HOLD THE
CAL SWITCH

CAL 0 IS DISPLAYEDENSURE PAN EMPTYPRESS ENTER WHEN
STABLEDISPLAYS SPN1 THENSCALE MAXIMUMUSE SAMPLE & SET
TO KEY IN INTENDED
CALIBRATION MASSPLACE MASS AND
PRESS ENTER
WHEN STABLEDISPLAYS ENDREMOVE MASS AND
PRESS CAL TOCOMPLETE

HW-G Series
52Simple Calibration Flow Chart
REMOVE COVER
AND PRESS CALSWITCH

LOWER DISPLA YSHOWS CALPRESS * KEY

LOWER DISPLA YSHOWS GRAVITYSETTING G9.XXX

PRESS C KEY

LOWER DISPLA YSHOWS CAL 0

LOWER DISPLA YSHOWS ------FOR 2 SECS.

ENSURE P AN
EMPTY ANDSTABLE PRESS
THE * KEY

XX REPRESENTS
THE SCALE MAX.
CAPACITY

LOWER DISPLA YSHOWS CAL XX

PLACE MASS ON
PAN AND WAIT
FOR STABLE

LOWER DISPLA YSHOWS ------FOR 2 SECS. AND
THEN CAL END

PRESS THE
* KEY

PRESS CAL KEY
TO COMPLETE

SF/SG Series
53Simple Calibration Flow Chart
Display off.

“CAL” displayed

While pressing the
RE-ZERO &UNITS key, press
the ON/OFF key.

“CAL 0” displayed

Press theRE-ZERO key

With no load on theweigh pan press the
RE-ZERO key“CAL F” displayed

Place test mass
onto the weigh pan.
Once stableindicator displayed
Press theRE-ZERO key“END” displayed

Scale automaticlly
returns to weighing
mode.

SJ-H Series
54Simple Calibration Flow Chart
DISABLE THE
AUTO POWER
OFF FUNCTION

WARM UP SCALE
FOR AT LEAST 10MINUTESTURN OFF USING
ON/OFF KEY

OLDER MODELSHIDDEN KEY IS BETWEEN RE-ZERO & ON/OFF
NEWER MODELSHIDDEN KEY IS TO THE LEFT OF RE-ZERO

ENSURE PAN EMPTYPRESS & HOLD REZERO & HIDDEN KEYDISPL AYS CALPRESS ON/OFFDISPL AYS CAL 0

PRESS RE-ZERO

CALIBRATES ZERODISPLAYS CAL FPRESS RE-ZERO

PLACE CAL MASSWAIT STABLE
PRESS RE-ZERODISPLAYS END

END

SK Series
55

Simple Calibration Flow Chart
DISABLE THE
AUTO POWER
OFF FUNCTION

WARM UP SCALE
FOR AT LEAST 10MINUTESCAL SWITCH IS LOCATED APPROX 4 CM
DEPTH BELOW COVER

INVERT SCALE &REMOVE CAL
SWITCH COVERDISPL AYS CALPRESS CALDISPL AYS CAL 0

PRESS RE-ZERO

CALIBRATES ZERODISPLAYS CAL FPRESS RE-ZERO

PLACE CAL MASSWAIT ST ABLE
PRESS RE-ZERODISPLAYS END

END

SK-WP Series
56Simple Calibration Flow Chart
WARM UP SCALE FOR
AT LEAST ONE HOUR

PRESS ON/OFF TO TURNDISPLAY OFF

PRESS & HOLD TARE

CLEARS STORES TARE
& ZERO VALUESPRESS ON/OFF

SW2-1 MOST EFFECT(ON INCREASES OFFSET)
VR4 FOR FINE ADJUSTUSE DIP SW2 & VR4 TO
ADJUST FOR ZERODISPLAY

PLACE MASS ON SCALE

SW1-1 MOST EFFECT(ON TO INCREASESREADING)
VR3 FOR FINE ADJUSTUSE DIP SW1 & VR3 TO
ADJUST FOR MASSVALUE DISPLAYREMOVE MASS &CHECK ZEROREMOVE MASS &CHECK ZERO READING

MAY NEED TO LOOPA FEW TIMES

END

AD-431657

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST ONE HOUR

PRESS STANDBY/OPERATE TO TURNDISPLAY OFFCAL DISPLAYEDSLIDE THE CAL SWITCH
ON (UP)

PRESS STANDBY/OPE RATEYESSPAN AT MAX?

NODISPLAYS CAL-1 FOR 3
SECONDSPRESS ZEROALL DECIMALS FOR 3
SECONDS THEN 6 ZEROSWITH FOURTH FLASHINGENSURE STABLE EMPTYSCALE & PRESSSTANDBY/OPERATECAL DISPLAYED FOR 3
SECONDS THEN BLANKLOAD CALIBRATIONMASS ONTO SCALE

ENSURE STABLE EMPTYSCALE & PRESSSTANDBY/ OPER ATE

USE ZERO, GROSS/NET
AND TARE KEYS

SET ACTUAL MASS

LOAD FULL CAPACITY
ONTO SCALE

7 DECIMALS DISPLA YEDPRESS STANDBY/OPE RATE

SWITCH OFF THE CAL
SWITCH

END

AD-432158

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST ONE HOUR

DISPLAY SHOWS d1

SLIDE CAL SWITCH ON(USE KEYBOARD)DISPLAYS d x

SET DIVISION (x) &PRESS ENTERDISPLAYS
CAP nnnnn

SET MAXIMUMCAPACITY (nnnnn) &PRESS ENTERCAL 0

PRESS ENTER TO CALZERO(PRESS TARE TOSKIP)DISPLAYS CAL SPn

PLACE CAL MASS ONSCALEDISPLAYS(CAL SPn m m m m m )SET CALIBRATION MASSIF NOT MAXIMUMDISPLAYS CAL END

PRESS ENTERSLIDE CAL SWITCH
OFF

END

AD-432259

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAY SHOWS CAL FOR
A FEW SECS. THENSHOWS d 01

SLIDE CAL SWITCH ON

USE NET/GROSS KEY TO
SELECT DIVISIONSET DIVISION & PRESSSTANDBY/OPE R AT EDISPLAYS CAP FOR A
FEW SECONDSFOLLOWED BY CO 10000

SET MAXIMUMCAPACITY & PRESSSTANDBY/OPE R AT E

USE ZERO; TARE; NET/GROSS; PRINT TO SETCAPACITYDISPLAYS PCALPRESS STANDBY/OPE RATE0DISPLAYS . . . . FOR A
FEW SECONDSFOLLOWED BY CAL SPn
FOLLOWED BY
XXXX - THE MAXIMUMCAPACITYSET CAL MASS TO BE
USED, IF NOT MAX.
PLACE MASS ON SCALEPRESS STANDBY/OPE RATEDISPLAYS ....... FOR A FEWSECONDS FOLLOWED BY
CAL 0

REMOVE ALL MASSFROM SCALE & PRESSSTANDBY/OPE R AT EDISPLAYS....... FOLLOWEDBY CAL END

SLIDE THE CAL SWITCH
OFF

END

AD-432360

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAY SHOWS
CAL
d 01

SLIDE CAL SWITCH ON

USE + & - KEYS TO
SELECT DIVISIONSET DIVISION & PRESSENTERDISPLAYS CAP CO 10000
USE +, -, > &< KEYS
TO SET MAXIMUM

SET MAXIMUMCAPACITY & PRESSENTERDISPLAYS PCAL 0

ENSURE SCALEUNLOADED & PRESSENTERDISPLAYS...... FOR A FEWSECONDS FOLLOWED BY
SPn 0 XXXX -THE
MAXIMUM CAPACITYSET CAL MASS TO BE
USED, IF NOT MAX.
PLACE MASS ON SCALEPRESS ENTERDISPLAYS....FOR A FEWSECONDS FOLLOWED BY
CAL 0

REMOVE ALL MASS FROMSCALE & PRESS ENTERDISPLAYS..... FOLLOWEDBY EndSLIDE THE CAL SWITCH
OFF

END

AD-4324
61

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAY SHOWS
CAL
d
01

SLIDE CAL SWITCH ON

USE + & - KEYS TO
SELECT DIVISIONSET DIVISION & PRESSENTERDISPLAYS CAP
C0 10000
USE +, -, > &< KEYS
TO SET MAXIMUM

SET MAXIMUMCAPACITY & PRESSENTERDISPLAYS PCAL 0

ENSURE SCALEUNLOADED & PRESSENTERDISPLAYS .... FOR A FEWSECONDS FOLLOWEDBY SPn o XXXXX - THE
MAXIMUM CAPACITYSET CAL MASS TO BE
USED, IF NOT MAX.
PLACE MASS ON SCALEPRESS ENTERDISPLAYS .... FOR A FEWSECONDS FOLLOWEDBY CAL 0

REMOVE ALL MASS FROMSCALE & PRESS ENTERDISPLAYS .... FOLLOWEDBY EndSLIDE THE CAL SWITCH
OFF

END

AD-4325
62

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 10 MINUTESDISPLAY SHOWS CAL FOR
2 SECONDS FOLLOWED
BY d-01 (CURRENT SETVALUE)PRESS CAL SWITCHUSE SAMPLE KEY ON"A" TYPE, KEYBOARD
ON "B" TYPESET DIVISION & PRESSENTERDISPLAYS CAP 10000
USE ARROW KEYS ON"A" TYPE, KEYBOARD
ON "B" TYPE

SET MAXIMUMCAPACITY & PRESSENTERDISPLAYS CAL 0

ENSURE SCALEUNLOADED & PRESSENTERDISPLAYS SPAn
FOLLOWED BY 1000
USE ARROW KEYS ON"A" TYPE, KEYBOARD ON"B" TYPESET CAL MASS TO BE
USED, IF NOT MAX.
PLACE MASS ON SCALEPRESS ENTERDISPLAYS C OFF

PRESS CAL SWITCHDISPLAYS End

PRESS ON/OFF TWICETO RETURN TO
WEIGHING MODE

END

AD-4326/7
63

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 10 MINUTESCAL DISPLAYED FOR A
FEW SECONDSFOLLOWED BY d-01

PRESS CAL SWITCHYES
CHANGEDIVISION?

SELECT DIVISION
WITH TARE BUTTONNODISPLAYS CAL-1 FOR
3 SECONDSPRESS PRINT/ENTER
TO STORE SELECTIONPRESS MODE BUTTONDISPLAYS CAPYESCHANGECAPACITY?
CHANGE USING TARE &ZERO BUTTONSNOPRESS MODE BUTTONPRESS PRINT/ENTER
TO STORE SELECTIONDISPLAYS CAL 0

PRESS MODE TO SKIP

ENSURE SCALE EMPTY &PRESS PRINT/ENTERDISPLAYS SPAn

SELECT SPAN MASSWITH TARE & ZEROPRESS MODE TO SKIP

PLACE MASS & PRESSPRINT/ENTERDISPLAYS END

PRESS CAL SWITCH
WHEN CALL OFFDISPLAYED

END

AD-432864

Simple Calibration Flow ChartDISPLAYS CAL in
FOR 2 SECONDS
THEN d-01

PRESS CAL KEY

SELECT REQUIREDDIVISION WITH
TARE KEYDISPLAYS MAX.
CAPACITY AS
C 10000

PRESS PRINT KEY

SET REQUIREDMAXIMUM WITH
TARE AND ZEROKEYS

PRESS PRINT KEYDISPLAYS CAL 0

ENSURE SCALEEMPTY AND NO MD
THEN PRESS PRINTDISPLAYS FXXXXXXXX=CAP ACITYSET CAL MASSWITH TARE AND
ZERO KEYS

PLACE MASS ON
SCALE AND PRESSPRINT WHEN NO
MOTIONDISPLAYS CAL End

PRESS CAL THEN
OPERATE/STANDBYTO COMPLETE

AD-432965

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESSEGMENT 6 ON 8 WAY
DIP ON EACH A/D CARDSWITCH OFF THE ZEROTRACKING

MOMENTARILY SHORTB10 TO B12 ON THE A/D
CARD I/O CONNECTOR

CLEAR ANY STOREDTARE ON THE A/D CARD

DIP SWITCH SEGMENT
1 HAS LEAST EFFECT.
FINAL ADJUSTMENT
WITH TRIMMERENSURE THE SCALE IS
EMPTY AND ADJUSTTHE ZERO READINGDIP SWITCH SEGMENT
1 HAS LEAST EFFECT.
FINAL ADJUSTMENT
WITH TRIMMERPLACE CAL MASS ON
SCALE & ADJUSTREADING TO MATCH

MAY NEED TO LOOPHERE AS SPAN HAS
EFFECT ON ZEROREMOVE CAL MASS &RE-CHECK ZEROSEGMENT 6 ON 8 WAY
DIP ON A/D CARDSWITCH ON ZEROTRACKING

REPEA T SEQUENCE FOR
EACH A/D CARD

END

AD-4335/35C
66

Simple Calibration Flow Chart
PRESS CALCAL SEtDISPLAYEDPRESS ENTERCAL 0
DISPLAYEDENSURE SCALEEMPTY THEN
PRESS ENTER----- DISPLAYEDFOR 2 SECS.

DISPLAYSCAL SPn

SUB DISPLAY
SHOWS CAPACITYSET CAL MASS
TO BE USEDPLACE MASSPRESS ENTER----- DISPLAYEDFOR 2 SECS.

DISPLAYSCAL ENd

PRESS ENTERTWICE TO EXIT

AD-4401
67

Simple Calibration Flow Chart
HOLD ENTER& PRESSDISPLAYS
THE MENU

USE
TO
SELECT CAL.
PRESS ENTERSHOWSCalibration mode
CAL
G

PRESS ENTERSHOWS
Set Unit
0:None 1:g 2:kg 3:t

SET THE REQUIREDUNIT & PRESS
ENTER

SHOWS
Set Decimal point. 0-4
0:None 1-4:Selected lights

SET THE REQUIREDDECIMAL PLACE
PRESS ENTER

1:1 2:2 3:5 4:10 5:20 6:50

SET THE REQUIREDDIVISIONPRESS ENTER

Set Capacity
16000

KEY IN CAPACITYPRESS ENTER

Zero calibration. Wait forstability W/O load.

Set Division

CONTINUED ON
NEXT PAGE

AD-4402
68

Simple Calibration Flow Chart
FROM
PREVIOUS
PAGE

SHOWS
Input weight mass value
16000

WHEN EMPTYSCALE IS ST ABLE
PRESS ENTERSHOWSSpan calibration. Wait forstability w/weight mass on.
16000

USE KEYS TOSET CAL MASSPRESS ENTER

PLACE CHOSEN
MASS ON SCALE
WAIT FOR STABLE
PRESS ENTERSHOWSCalibration complete

REMOVE CAL MASSSHOWS ZERO ON MAINDISPLAYPRESS ESCSEVERAL TIMESTO RETURN
TO WEIGHING

END
OF
PROCEDURE

AD-4402
69

Simple Calibration Flow Chart
WARM UP SCALE
FOR AT LEAST 30
MINUTESDISPLAYS THE MENU IN STOP MODE

PRESS & HOLDENTER THEN
PRESS

PRESS
TO
SELECT CAL &PRESS ENTERSKIPS THE WEIGH UNIT, DECIMAL POINT,MIN DIV & MAX CAPACITY SETTINGS
SEE HANDBOOK IF YOU NEED CHANGESPRESS ENTER 4TIMESENSURE EMPTY
PAN & ST ABLE
PRESS ENTER

USE THENUMERICALKEYS

SET CAL MASSVALUEPRESS ENTER

PLACE MASSWAIT STABLE
PRESS ENTERESC SEVERALTIMES TO STOPMODEPRESS ESC

END

AD-4404
70

Simple Calibration Flow Chart
WARM UP SCALE
FOR AT LEAST 30
MINUTESDISPLAYS CAL IN FOR 2 SECONDS
THEN DISPLAYS CAL 0

REMOVE COVER& PRESS CALCALIBRATES ZERO,DISPLAYS CAL F
THEN FULL CAPACITY MASSENSURE P AN
EMPTY & STABLE
PRESS ENTER

USE THE NUMERIC KEYS TO
SET THE REQUIREDCAL MASS

IF NECESSARYSET CAL MASS
TO BE USEDDISPLAYS CALEnd

PLACE MASSWAIT STABLE
PRESS ENTERCALIBRATION DATASTOREDDISPLAYS CALoFF

PRESS CALPRESS ON/OFF

END

AD-4405
71

Simple Calibration Flow Chart
WARM UP SCALE
FOR AT LEAST 30
MINUTESDISPLAYS CAL IN FOR 2 SECONDS
THEN DISPLAYS CAL 0

REMOVE COVER& PRESS CALCALIBRATES ZERO, DISPLAYS CAL F
THEN FULL CAPACITY MASSENSURE P AN
EMPTY & STABLE
PRESS ENTER

SELECT DIGIT WITH KEY
SET VALUE WITH KEY

IF NECESSARYSET CAL MASS
TO BE USEDDISPLAYS CALEnd

PLACE MASSWAIT STABLE
PRESS ENTERCALIBRATIONDATA STOREDDISPLAYS End

PRESS CALPRESS ON/OFF

END

AD-4406
72

Simple Calibration Flow Chart
WARM UP SCALE
FOR AT LEAST 30
MINUTESDISPLAYS CAL IN FOR 2 SECONDS
THEN DISPLAYS CAL 0

REMOVE CO VER& PRESS CALCALIBRATES ZERO, DISPLAYS CAL F
THEN FULL CAPACITY MASSENSURE PANEMPTY & STABLE
PRESS ENTER

USE THE NUMERIC KEYS TO
SET THE REQUIREDCAL MASS

IF NECESSARYSET CAL MASS
TO BE USEDDISPLAYS CALEnd

PLACE MASSWAIT STABLE
PRESS ENTERCALIBRATION DATASTOREDDISPLAYS CALoFF

PRESS CALPRESS ON/OFF

END

AD-440773

Simple Calibration Flow Chart
Warm up scale forat least 30 minutes

Displays CAL SET

Remove cover &press CAL
After CAL is displayed,press ENTER

Displays CAL 0

Press ENTER

Calibrates ZERO, thendisplayes CAL SPn

Ensure pan
empty & STABLE
Press ENTER

Press ENTER
If necessary, set cal
mass to be used

Displays CAL End

Place mass
Wait STABLE
Press ENTER

Calibration data stored
Displays CAL SEt

Press ESC

Press ON/OFFEnd

AD-4408A/C
74

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAYS FunC

HOLD LEFT ARROWKEY DEPRESSED THEN
PRESS ENTER FORTWO SECONDSDISPLAYS CALPRESS LEFT ARROWKEY TWICEDISPLAYS CAL 0

PRESS ENTERDISPLAYS .... WHILE ZEROCAL PERFORMED THENDISPLAYS SPAn

ENSURE SCALE EMPTY& PRESS ENTER

USE LEFT & UP ARROWKEYS

SET SPAN CAL MASS &PLACE MASS ON SCALEDISPLAYS CAL MASS
FOR A FEW SECONDS& THEN DISPLAYS End

PRESS ENTER

UNIT REVERTS TONORMAL WEIGHING

END

AD-4531/2
75

Simple Calibration Flow Chart
Warm up scale forat least 30 minutes

Displays Fnc

Press FNC key
for 3 seconds

Displays CAL

Press

Performs zero
calibration

key

Ensure scale empty& press
key

Set span cal mass &place mass on scale

Displays cal mass for
2 seconds & thendisplays End

Wait STABLE
Press ENTER

Press ESCEnd

AD-4532B
76

Simple Calibration Flow Chart
PRESS & HOLD
PROGRAM UNTILCHDISPLAYS CHPRESS SELECTDISPLAYS CALPRESS ENTERDISPLAYS CAL 0

PRESS ENTERDISPLAYS 20

PLACE MASS (20g)
AND PRESS ENTERDISPLAYS End

RETURNS TOREMOVE MASSWEIGH MODE

END

M Series Moisture Analyzers
77Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESREMOVE THE WEIGHINGPA N

DISPLAYS TARE KEY

PRESS & HOLD THE UP
KEY, RESET KEY & TAREKEY TOGETHER FOR A
FEW SECONDSDISPLAYS CAL 100g

PRESS TARE

PLACE 100g MASS ON
CENTRE OF TRIVET

WAIT FOR 5 SECONDSMINIMUM AND THEN
PRESS TAREDISPLAYS BUSYREMOVE THE 100g CAL
MASS & REPLACE PAN

PRESS TARE. UNITREADY FOR MOISTUREANALYSIS

END

AD-471278

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESDISPLAYS CALIBRATIONPRESS & HOLD THE UP
KEY, RESET KEY & TAREKEY TOGETHER FOR A
FEW SECONDSDISPLAYS CAL 200g

PRESS TARE

PLACE 200g MASS ON
PA N

WAIT FOR 5 SECONDSMINIMUM AND THEN
PRESS TAREDISPLAYS CAL END
FOLLOWED BY BUSYREMOVE THE 200g CAL
MASS & REPLACE PAN

PRESS TARE, UNITREADY FOR MOISTUREANALYSIS

END

AD-471379

Simple Calibration Flow Chart
WARM UP SCALE FOR AT
LEAST 30 MINUTESREMOVE THE WEIGHING
PAN & TURN THE
BALANCE OFFDISPLAYS BUSYFOLLOWED BY CALMODE uX.X

PRESS & HOLD THE
UP ARROW KEY AND
TURN ON THE POWER
SWITCH. CONTINUETO HOLD THE KEY FOR
SEVERAL SECONDSDISPLAYS REMOVEPLATE FOLLOWED BYCAL ZERO PUSH KEY

RELEASE THE DOWN
ARROW KEYDISPLAYS PUT 100g
PUSH KEY

PRESS TARE KEYDISPLAYS CAL END
AFTER A FEW SECONDS.
THEN DISPLAYS SET
PLATEPLACE 100g MASS ON
PAN & PRESS TAREREMOVE THE 100g CAL
MASS & REPLACE PANDISPLAYS 0.00g & TIME
& TEMPERATURE
SETTINGSPRESS START

END

AD-471480

Simple Calibration Flow Chart
ALLOW 1 HOUR
WARM UP TIMEDISPLAY SHOWSCAL 0:200.000g

PERFORMS ZEROPRESS & HOLDENTER & THEN
PRESS TARECONFIRM PANCALIBRATIONEMPTY, CLOSEDISPLAYS

COVER & THEN

>CAL0:200.000g

PRESS SELECTENSURE THE

DISPLAYS
CAL. W:200.000g

PLACE 2 x 100g

MASSES WILL NOT

PRESS SELECT BEFOREMASSES ON PANTOUCH THE SENSORPLACING MASSES IF ONLY

AND CLOSE COVER

OR GLOBE WHEN
COVER CLOSEDZERO CAL REQUIREDDISPLAYS>CAL W:200.000g

PRESS SELECT TOPERFORMS SPAN

CALIBRATE SPAN

CALIBRATIONDISPLAYSCALIBRATION ENDREMOVE MASSES

E ND

AD-471581

Simple Calibration Flow Chart
CAL ISDISPLAYEDPRESS & HOLD
THE HOLD KEY

CAL - 1IS DISPLAYEDPRESS MODE TO
SELECT ONE
POINT CALPRESS PRINTPLACE STANDARD
VISCOSITY FLUIDIN SAMPLE CUPMEASUREMENTSTARTSPRESS STARTWAIT FORSTABLE DISPLAYPRESS PRINTVALUE BLINKSREADY FOR
CORRECTION

USE MODE,START, STOP TO
CORRECT VALUEPRESS PRINT TOCONFIRMEndIS DISPLAYEDPRESS PRINT

END

SV/SV-A Series
82Simple Calibration Flow Chart
Simplified Calibration(For the SV-10/1A/10A Only)

MEASURES THE PURIFIED WATERPLACE PURIFIEDWATER IN THE
SAMPLE CUPPRESS STARTENSUREVISCOSITY & TEMPARE STABLE

THEORETICAL VISCOSITYVALUE DISPLAYED AND ALLDISPLAYS BLINK

PRESS & HOLDSTARTCALIBRATIONPERFORMEDPRESS STARTAGAIN

End DISPLAYEDON COMPLETION

RETURNS TO
MEASURINGSTATUS

END

SV-10/1A/10A
83