Product Datasheet

Optifit Tips
Standard Multipurpose
Tips for Various Applications

Product Information

Description

Sartorius Optifit Tips are ideal for use with for Sartorius
pipettes, enabling maximum accuracy and precision, as
well as comfortable ergonomics. Sartorius pipettes
feature a unique spring-loaded Optiload mechanism thatmakes it easy to attach and eject tips while still ensuring aperfect seal. Non-filter Optifit Tips are available in various
packaging types, purity certified and pre-sterilized.

Optifit Tips are manufactured from virgin polypropylene.
Strict quality standards (ISO 9001 and ISO 13485) are followed at every step of the process, from R&D to production
and delivery. The tips are manufactured in ISO 8 classified
cleanroom conditions to avoid any contamination and are
certified and tested for DNase, RNase, human DNA and endotoxins. Presterilized tips are also available for laboratory
work where purity requirements are of critical importance.
The tips are released for sale once they have fulfilled the test
requirements.
Certificates are available at www.sartorius.com.

-----Features

2

Available in volumes from 10 μl to 10 ml including
extended and wide-bore tips
Products in single tray, refill pack and FlexiBulk® packages are certified as DNase, RNase, human DNA andendotoxin free
Informative rack labels indicating tip volume, productnumber, lot number, expiration date, and purity level for
improved tip identification and traceability
E-beam pre-sterilized tips available for sterile work
Perfect fitting and tip sealing with all Sartorius pipettes
Smooth and ergonomic tip attachment and ejection
ensured by compatibility with the Optiload featureof Sartorius pipettes
Color-coded tip trays make it easy to match tip withappropriate Sartorius pipette
Flip-top racks that are easy to open and close with
one handMaterials used in tips, trays and racks are 100 % recyclable
or can be incinerated as energy waste-Tip racks can be refilled with tips from Bulk or Refill
packages. The racks and tips are fully autoclavable(121 ºC, 20 min, 1 bar).

The Sartorius tip offering also includes Safetyspace™ Filter
Tips (see catalog or datasheet for details).

Intended Use
The Sartorius Optifit Tips are intended, designed andmanufactured to be used in liquid dispensing in a variety ofapplications and to be used in combination with Sartorius
pipettes.
It is recommended that Optifit Tips are used with Sartorius
pipettes to ensure optimum compatibility and performance.

Technical Data
Materials and Recylability

Technical Specifications*
Sterilization

Purity certificate

Endotoxin purity testing

DNase purity testingElectron beam (beta irradiation) in accordance with EN 552 and ISO 11137 andvalidated by using 25 kGy as the minimum
dose with SAL (sterility assurance level)of 10-6.
Tips are tested and certified to be free of
DNase, RNase, human DNA andendotoxins
Validated test result with Chromogenic
kinetic method according to Ph.Eur. 2.6.14,
Method D for endotoxin free pipette tip is< 0.005 IU/mL (EU/mL).
DNase-free pipette tips show no evidenceof DNase activity in the fluorometric assay
with the detection level of < 6.25*10-5 U/μL
when DNase I is used as a standard.

RNase purity testing

RNase-free pipette tips show no evidenceof RNase activity in the fluorometric assay
with the detection level of <1*10-8 U/µl.

Human DNA purity testing

Human DNA free pipette tips show no evidence of human DNA in quantitative PCR(qPCR) using human DNA specific primers
and with a detection level of < 1 pg/μL.

Standards

ISO 9001 Quality management standard
ISO 14001 Environmental standard
ISO 13485 Medical device quality
management standard
ISO 14644-1 Class 8 cleanroom

Autoclavability

Tips, trays, and racks are autoclavable(121 °C, 20 min, 1 bar)

Transportation
conditions

Temperature: –50 °C to +50 °C
Humidity: 10 % – 90 %Shocks: < 10 G

Storage conditions

Storage in room temperature isrecommended
Temperature: –20 °C to +40 °C
Humidity: 10 % – 60 %Expiry date

5 years from manufacturing date** Extended filter tip sterility assurance level, DNase, RNase, human DNA and endotoxin testing
detection limits may differ from values reported here. You can find batch specific certificates at
www.sartorius.com

Tip

Polypropylene (PP)
Recyclable, can be incinerated asenergy waste

Tray and Rack

Polypropylene (PP)
Recyclable, can be incinerated asenergy waste

FlexiBulk® pack

Polyethylene terephthalate (APET) /Polyethylene (PE)Recyclable, can be incinerated asenergy waste

Cardboard package

Cardboard
Recyclable, renewable waste

Package Options

FlexiBulk® Pack

Single Tray

Refill Pack

Bulk packaging

Refill Tower
3Ordering Information
Optifit Tips
Order CodeVolume Range

LengthPackaging

Purity Level
Free of DNase, RNase,human DNA & endotoxins

79001079001179001279001379001410 µL

31.5 mm

Single Tray
Single Tray
Refill Tower
Refill Pack
Bulk in Bag

790200790201790202790203
LH-B790204

200 µL

51 mm

Single Tray
Single Tray
Refill Tower
Refill Pack
FlexiBulk®

790350790351790352790353
LH-B790354

350 µL

54 mm

Single Tray
Single Tray
Refill Tower
Refill Pack
FlexiBulk®

791 000
791001791002791003
LH-B791004

1,000 µL

71.5 mm

Single Tray
Single Tray
Refill Pack
Refill Pack
FlexiBulk®

791020791021
LH-B791024 1,000 µL
Wide bore tip

68.5 mm

Single Tray
Single Tray
FlexiBulk®

791 200
791201791202791203
LH-B791204

1,200 µL

71.5 mm

Single Tray
Single Tray
Refill Pack
Refill Pack
FlexiBulk®

7803047803057803007803085,000 µL

150 mmSingle Tray
Single Tray
Bulk in BagBulk in Bag

LH-780314
LH-780316

10,000 µL

155 mmSingle Tray
Bulk in Bag

4-------------

Tips/Unit
Pre-sterilized-10 × 96
10 × 96
10 × 96
20 × 96
1,000
10 × 96
10 × 96
10 × 96
15 × 96960
10 × 96
10 × 96
10 × 96
15 × 96960
10 × 96
10 × 96
10 × 96
10 × 96480
10 × 96
10 × 96480
10 × 96
10 × 96
10 × 96
10 × 96
480
5050
100
1,00035250

Extended Standard Tips
Order CodeVolume Range

LengthPackaging

Purity Level

Tips/Unit

Free of DNase, RNase,human DNA & endotoxins783210
783211

10 µL

46 mm

Single Tray
Single Tray

LH-X780200
LH-X780201

200 µL77.5 mm

Single Tray
Single Tray

LH-X791000
LH-X791001

1,000 µL

119 mmSingle Tray
Single Tray

791210791211791212791213

1,200 µL

90 mm

Single Tray
Single Tray
Refill Pack
Refill Pack

Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0F
 or further information, visit
www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 10 | 2024------

USA
Sartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906

Pre-sterilized-10 × 96
10 × 96
10 × 96
10 × 96
10 × 96
10 × 96
10 × 96
10 × 96
10 × 96
10 × 96

Finland & Baltics
Sartorius Liquid Handling Oy
Tulppatie 100880 Helsinki
lhinfo.finland@sartorius.com