Product Datasheet

Prospenser andProspenser Plus
Bottle-topDispensers
Product InformationFeatures

Prospenser and Prospenser Plus arehigh quality bottle-top dispensers withexcellent chemical resistance. They aresafe, convenient, and customizable,
making them perfect for a wide rangeof dispensing needs.

 Extremely high chemical and thermal resistance due to a ceramic piston,glass cylinder, and highly resistant fluoroplastic parts
 The incremental volume setting ensures accurate and repeatable results Smooth and light to operate
 Easy to attach aspiration tube with a secure threaded connection
 User-replaceable valves Wide range of accessories
 Fully autoclavable

Description | Intended Use
Prospenser and Prospenser Plus are
premium bottle-top dispensers available
in a volume range from 0.2 to 60 mL,
used for manual dispensing of liquids.

Additional Prospenser Plus Features
 Media recirculation system that allows unused media to be recoveredfrom the device and returned to the bottle 360° rotating dispensing head
 Universal connector at the back of the unit for drying tubes and filters

Technical DataWeight200 g (5 mL, Prospenser) | 388 g (5 mL, Prospenser Plus)

Length143 mm (5 mL, Prospenser) | 175 mm (5 mL, Prospenser Plus)

Autoclavability

Fully autoclavable (121 ºC, 20 min, 1 bar/100 kPa)

Threaded bottle fitting

Prospenser: A32 | Prospenser Plus: A45

Bottle adapters included

Prospenser: A28, S40, A45 | Prospenser Plus: A32, A38, S40

Limitations

The accuracy and precision values are valid under the controlled conditions described in ISO 8655.
Test temperature: 15 ºC to 30 ºC +/-0.5 ºC. Relative humidity: > 50%

Transportation conditions

Temperature: 0 ºC to 40 ºC

Long-term storage conditions

Temperature: 0 ºC to 40 ºC

Chemical resistance

Information available on request

Materials

Piston: Highly pure AI2O3-ceramic (99.7%) | Cylinder: Borosilicate glass 3.3 | Valve block: Halar (ECTFE)
Valve ball: Borosilicate glass 3.3 | Valve seat: Borosilicate glass 3.3 | Valve spring: Hastelloy (99.7%)

Warranty

2 years

Ordering Information
Propenser

Propenser
Plus

VolumeRange (mL)

Increment(mL)

Test Volume(mL)

Systematic
Error* (%)

SystematicError* (mL)

RandomError* (%)

RandomError* (mL)

LH-723060

LH-723070

0.2--1

0.05

10.60.006

0.20.002

LH-723061

LH-723071

0.4--2

0.05

20.6

0.012

0.20.004

LH-723062

LH-723072

1--5

0.10

50.6

0.03

0.20.01

LH-723063

LH-723073

2--10

0.25

10

0.6

0.06

0.20.02

LH-723064

LH-723074

5--30

0.50

30

0.6

0.18

0.20.06

LH-723065

LH-723075

10--60

1.00

60

0.6

0.36

0.20.12* The listed systematic and random error values can be achieved under strictly controlled conditions during type tests as per ISO 8655.
Due to our continuous product development, the systematic and random error values may change without prior notice.

2YearWarranty

Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Göttingen
Phone +49 551 308 0

For additional information,
visit www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 07 | 2022

2USA
Sartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906

2YearWarrant