Digital Blood Pressure Monitor
Model UA-789XL
Instruction Manual
Original

Manuel d’instructions
Traduction

Manual de Instrucciones
Traducción

Manuale di Istruzioni
Traduzione

BedienungsanleitungÜbersetzung使用手冊
翻譯
1WMPD4001765C

目 錄
給親愛的顧客......................................................................................................................................................................................................................................................1
前言............................................................................................................................................................................................................................................................................2
用前需知..................................................................................................................................................................................................................................................................2
用件說明..................................................................................................................................................................................................................................................................3
標識含意..................................................................................................................................................................................................................................................................4
操作方式..................................................................................................................................................................................................................................................................5
使用控制器............................................................................................................................................................................................................................................................5
安裝／更換電池......................................................................................................................................................................................................................................5連接充氣管子...........................................................................................................................................................................................................................................6連接交流電轉接器................................................................................................................................................................................................................................6
調整內裝的時鐘......................................................................................................................................................................................................................................6
設定三個預警告裝置..........................................................................................................................................................................................................................7
顯示屏預警開關(ON/OFF)標識....................................................................................................................................................................................................7
選擇適當的袖帶......................................................................................................................................................................................................................................7繞緊袖帶於上臂......................................................................................................................................................................................................................................8
怎樣正確測量...........................................................................................................................................................................................................................................8
測量..................................................................................................................................................................................................................................................................8
測量結束後.................................................................................................................................................................................................................................................8
測量方法..................................................................................................................................................................................................................................................................9
怎樣正確測量血壓................................................................................................................................................................................................................................9
以合理的收縮壓測量您的血壓.................................................................................................................................................................................................10
注意正確的測量方法........................................................................................................................................................................................................................10查詢數據................................................................................................................................................................................................................................................................10什麼是不整脈....................................................................................................................................................................................................................................................11
血壓柱指示器....................................................................................................................................................................................................................................................11
WHO分類指示器...........................................................................................................................................................................................................................................11有關血壓的一些知識...................................................................................................................................................................................................................................11
什麼是血壓？.........................................................................................................................................................................................................................................11
什麼是高血壓？它是怎樣被控制的？................................................................................................................................................................................12
為什麼在家裡測量血壓？.............................................................................................................................................................................................................12WHO(世界衛生組織)血壓分類標準.......................................................................................................................................................................................12
血壓差異.....................................................................................................................................................................................................................................................12故障檢修................................................................................................................................................................................................................................................................13
維修和保養..........................................................................................................................................................................................................................................................13
技術資料................................................................................................................................................................................................................................................................13

給親愛的顧客銘謝惠顧最新型的A&D血壓計。為追求本血壓計的正確利用和方便使用起見,使它能成為您每天血壓管理上的最佳助手。
敬請在使用前, 能詳細閱讀後使用。

中文 1

前言
本血壓計符合歐盟管理文件93/43EEC的有關醫療產品之相關規定﹐且有關權威機構認定的標記為證。(0123：有關通知本身的參照號碼。)
本血壓計只適用於成年人, 而非新生兒或嬰兒。
應用環境﹕本血壓計適用於家庭保健環境。
本血壓計專用於測量血壓與脈搏次數，供診斷用。

用前需知
本血壓計係專用於粗大手臂，並非為搭配中小型袖帶使用而設計。請搭配大型或超大型袖帶進行使用。
上臂尺寸

推薦袖帶尺寸訂購編號

32 cm 到 45 cm

大型袖帶

42 cm 到 60 cm

超大型袖帶

CUF-D-LA
CUF-B-XLA

本血壓計是採用精密部件製造的。盡量避免下述因素的過度異常﹐ 諸如溫度﹑ 濕度﹑ 陽光直射﹑ 震盪或灰塵。
用乾軟布，或是蘸水和中性洗滌劑的布清潔血壓計及袖帶。切勿使用酒精、苯、稀釋劑或其他刺激性的化學製品來清潔血壓計或袖帶。
避免過度用力折疊袖帶或儲存過度扭曲的軟管﹐ 放置很長時間。因為這樣會縮短相關設備的壽命。
本血壓計和袖帶不耐水﹐ 因此應避免淋雨汗水和水。
假若血壓計使用時靠近電視﹑ 微波爐﹑ 脈沖電話機﹑ Ｘ– 線或其它強電場時﹐ 其測量將會模
糊不準確。
用過的設備﹑ 零部件和電池不能作為一般家庭廢物處理﹐ 而應當按照相關的規定進行處理。
使用交流電轉接器時，請確認交流電轉接器可在必要時隨時從插座拔下。
再次使用血壓計時，請確認血壓計是乾淨的。
請勿改造血壓計，否則可能造成意外，或損壞血壓計。
若要測量血壓，袖帶必須要適當束緊手臂，暫停血液通過動脈。這可能會造成疼痛、麻木或手臂出現暫時性紅印。此狀況特別會在連續測量時出現。
任何疼痛、麻木或暫時性紅印將會隨時間消失。

禁忌
以下是正確使用血壓計的注意事項。
請勿將袖帶套到還縛上其他醫療電氣設備的的手臂，否則設備可能無法正常運作。
手臂血液循環不良的病患，請務必在使用血壓計前諮詢醫生，避免醫療糾紛。
請勿自我診斷測量結果，並開始自我治療。請務必向醫生諮詢結果評估與治療。
請勿將袖帶套到有傷口的手臂上。
請勿將袖帶套到接受靜脈滴注或輸血的手臂，否則可能造成傷害或意外。
請勿在有易燃氣體的地方使用血壓計，例如麻醉氣體，否則可能會導致爆炸。
請勿在高度濃縮的氧氣環境中使用血壓計，例如一個高壓氧氣室或氧帳，否則可能引起火災或爆炸。

中文 2

用件說明
儲存按鈕

顯示屏直流插座
開關按鈕 藍光時鐘按鈕充氣管子

袖 帶（特大）
電池蓋
充氣管插座

電池盒

充氣管連接插頭

1.5V電 池(R6P,LR6或AA)顯示器的名稱

存儲器

收縮壓舒張壓

血壓柱指示器 和WHO分類指示器

脈搏率電池指示器

心臟標識I.H.B.指示器(不整脈指示器)

中文 3

標識含意
功能 / 意義標識處置開關機構時鐘按鈕標識假若您想停止,按下START(開關)按鈕。

響鈴開關標識SYS.

收縮壓以mmHg表示

DIA.

舒張壓以mmHg表示

PUL./min.

脈搏每分鍾速率
裝填電池的指示方向
直流電BF型: 血壓計, 袖帶和相應的管子 已被設計對電震動有特殊保護作用。
當進行測量或測到脈搏時即顯示本圖像。
繼續測量﹐保持安靜。
在測量過程中,當測到心臟跳動不規律或身體過渡移動時即顯示本圖像。
前次測量的結果儲存於此Full Battery 在測量時電池電壓顯示器即顯示本圖像。
Low Battery 電池電壓太低時閃爍,即顯示本圖像。

在測量過程中由於身體移動導致血壓值不穩定時出現本標記。
當測量收縮壓和舒張壓時,都在10mmHg內時顯示本圖像。
袖帶加壓時﹐血壓值不增加時顯示本標記。
當袖帶不能正確拉緊時
PUL. DISPLAY
ERROR

當脈搏不能被正確測到時
EC管理機構認可的醫用儀器標識EU代表生產商
生產日期
ClassII裝置
WEEE標識參閱說明手冊SN系列號碼
直流插座極性

中文 4

更換新電池。
再次測量并且在測量時保持安靜。

正確拉緊袖帶并再次測量。

操作方式
1. 平時測量的保存數據(請參照9頁)
請按壓 START (開關)按鈕。開始測量, 同時血壓計開始記錄保存內容。
本血壓計的各按鈕能保存最近的數據60個。
請按壓儲存按鈕。

2. 查詢數據(請參照10頁)
請按壓儲存按鈕。日期和最新數據將會顯示。
本血壓計的各按鈕能保存最近的數據60個。

日期

月日時間

3. 刪除數據(請參照10頁)
壓住儲存按鈕直至“＂標識出現, 即為清除原儲存的數據。

數據4. 用希望的預期收縮壓測量
有關希望用預期收縮壓進行測量時, 請參照10頁。

使用控制器
安裝／更換電池
1. 打開電池蓋之前,先滑動電池蓋。
步驟22. 取出舊電池﹐裝入新的電池時, 要留意
電極(＋﹐－) 插入新電池（如圖所示）。
3. 未關上電池蓋之前,先滑動電池蓋。
僅限於使用R6P,LR6或AA電池。

步驟3步驟1注意
插入電池時﹐要按照電池盒内的電極方向,否則血壓計不能工作。
當（低電壓標記）顯示時﹐同時要用新電池更換全部舊電池﹐不要新舊電池一起混用。
當電池用盡時,（低電壓標記）並不會出現。
電池壽命與環境溫度有關﹐在低溫情況下﹐壽命縮短。
請使用指定的電池。與血壓計同時供應的電池是測試用的,壽命有限。
當血壓計長時間不用時應取出電池。電池液體的漏出是造成故障的原因。
取下電池時,儲存的數據資料就無法保存下來。

中文 5

使用控制器
連接充氣管子將空氣連接插頭緊緊插入空氣插座內。

充氣管連接插頭

連接交流電轉接器
將交流電轉接器插頭插入直流插座。
之後將交流電轉接器插入電源插座。
請使用指定的交流電轉接器。
(請參照第14頁。)

充氣管插座直流插座

交流電轉接器插調整內裝的時鐘本血壓計內裝有時鐘, 我們建議您在使用之前先調整時間。

時鐘按鈕

1. 按壓時鐘按鈕直到數字出現。
2. 用 START (開關)按鈕選擇年份。
按壓時鐘按鈕設定現在的年度,再選擇月／日。選擇期間為2004~2053年。
3. 用 START (開關)按鈕選擇月份。
按壓時鐘按鈕設定現在的月份,再選擇日期。

年月/日

4. 用 START (開關)按鈕選擇日期。
按壓時鐘按鈕設定現在的日期,再選擇時／分。
5. 用 START (開關)按鈕選擇小時。
按壓時鐘按鈕設定現在的小時,再選擇分鍾。
6. 用 START (開關)按鈕選擇分鍾。
按壓時鐘按鈕設定現在的分鍾數,再開始時鐘讀數。

中文 6時/分時鐘使用控制器
設定三個預警告裝置
本血壓計有三個預警告裝置,您可以設定24小時之內不同預警裝置。

時鐘按鈕

1. 壓著時鐘按鈕直至數字顯示出來,然後按壓儲存按鈕。此時”P-1”在顯示屏中間出現。

儲存按鈕

2. 應用 START (開關)按鈕選擇小時。
按壓時鐘按鈕設定小時, 然後再選擇 分鍾, 之後
按壓儲存按鈕即完成預警設置。

第一預警

3. 然後,顯示屏顯示”P-2”為第二個預警設定。
如果您需要第二個預警,即可重復用上述同樣方法。

第二預警
第三預警

4. 第三個預警裝置設定方法也一樣。

顯示屏預警開關(ON/OFF)標識當打開預警裝置開關時, “ ” 即出現在顯示屏上。當要關掉預警裝置時只需簡單再按一下時鐘按鈕。 預警裝置通常響一分鍾。關掉預警裝置只要再按一下 START (開關)即可。

選擇適當的袖帶
合適的袖帶尺寸對正確的測量很重要。袖帶大小不合適時,有可能測量不正確。
上臂尺寸標示於每條袖帶上。
袖帶屬於消耗品。如已不堪使用，請購買一個新袖帶。
上臂尺寸

推薦袖帶尺寸目錄編號

32 cm 到 45 cm

大型袖帶

42 cm 到 60 cm

超大型袖帶

CUF-D-LA
CUF-B-XLA

上臂尺寸:指肩到肘腕之間的上臂外圍尺寸。
注意:本血壓計UA-789 XL並未設計有小號的袖帶。

印在袖帶上的標識
標識

功能/意義動脈位置標識

REF

建議動作
請將

標識置於上臂動脈，或與無名指呈一直線的手臂內側。

目錄編號批號

中文 7

使用控制器繞緊袖帶於上臂

請勿過度緊繞袖帶

1. 纏繞袖帶於上臂﹐大約離肘部1～2cm 如圖所示。
因為衣服可引起脈博減弱導致測量誤差﹐請正確纏繞袖帶於上臂上。

袖帶
1～2cm

2. 過渡纏繞袖帶以致上臂收縮過緊﹐會影響血壓值的正確讀數。
動脈位置標識確保黏扣帶已
緊緊纏繞.

怎樣正確測量為了準確的測量血壓﹐應注意下列幾點。
放鬆坐好。將手臂放在桌上。請勿翹腳，兩腳平放坐直。
測量前先放鬆。
將袖帶的中心放置於相當與您的心臟同一高度。
測量時保持安靜。
體育活動或洗澡後不要馬上測量。請先休息後再測量。
儘量在每天的同一時間測量血壓

測量
在測量血壓時﹐通常要感到袖帶很緊方可(請不要驚慌)

測量結束後測量結束後﹐按壓 START (開關)按鈕關掉電源。取下袖帶並記錄測量數據。
註﹕ 本血壓計有自動電源關閉功能。在測量結束3分鍾後可自行關掉電源。

中文 8充氣管子

測量方法
本血壓計UA-789XL被設計用來測量脈博和使袖帶自動充氣到收縮壓水平。
如果您的收縮壓預計超過230mmHg﹐請參考下頁的“以合理的收縮壓測量您的血壓＂。

怎樣正確測量血壓
1. 將袖帶放於上臂(最好是左上臂),測量時保持安靜。
2. 按壓 START (開關)按鈕。
顯示出平均血壓和最近測量的簡略數據代碼。
之後開始測量, 如右圖所示顯示器開始變化。
袖帶開始膨脹。袖帶感到有些緊張這是正常的。
在袖帶膨脹的時候血壓柱指示器如右圖所示。

與心臟同高度START 按鈕START平均數據數據代碼註: 假若您想隨時停止充氣﹐只需再次按壓 START(開關) 按鈕。
開始膨脹
3. 當充氣完成時﹐放氣自動開始﹐且(心臟標記)閃爍﹐表明測量在進行中。
一旦測到脈搏﹐心臟標記將隨著脈搏的跳動而閃爍。
加壓註: 假若充氣不完全或不能得到合適的壓力﹐血壓計將自動再次充氣。
4. 當測量結束後﹐收縮壓和舒張壓讀數及脈搏率在顯示屏上顯示。袖帶完全排出剩余的空氣。
開始測量
5. 再按壓 START (開關)按鈕關掉電源。
收縮壓

註: 本血壓計UA-789XL有自動電源關閉功能﹐在測量結束一分鍾後﹐可自行關掉電源。

舒張壓WHO分類
脈搏率自動排除剩余的空氣

中文 9

測量方法
以合理的收縮壓測量您的血壓如果收縮壓預計超過230mmHg,請按下面步驟進行。

與心臟同高度1. 將袖帶放於上臂(最好是左上臂)。
2. 壓住 START (開關)按鈕直到比您的預測的血壓高30~40mmHg的數字出現。

按住 START (開關)按鈕

3. 當預測的數字出現時,放開 START (開關)按鈕,測量即自動開始,然後參考前頁繼續測量您的血壓。

當預計的收縮壓出現時放開開關按鈕
測量方法參考前頁。

注意正確的測量方法
舒適而又安靜地坐下,把通常要測量的手臂放在桌子或其它物體上面﹐以便讓袖帶中心和心臟處
於同一水平高度。
開始測量前輕鬆一下大約5～10分鐘。假若由於情緒關係過於激動或壓抑﹐則測量值將比正常血壓高或低而脈博比正常要快。
一個人的血壓值通常會有些變化﹐這些取決於您當時在做什麼和當時的飲食,飲酒對您的血壓有強而又有效的影響。
本血壓計根據您的心臟鼓動而測量血壓。假若您的心臟鼓動微弱或不規律時, 血壓計很難測到您的血壓。
假若血壓計檢測狀況異常, 它將會停止測量並顯示錯誤符號。具體細節請參考第4頁標識含意的說明。
本血壓計僅適用於成人, 孩童使用前請先諮詢醫師建議。無人陪伴的孩童不應使用本血壓計。

查詢數據
註: 本血壓計自動儲存前六十次的測量數據﹐只要電池一直在血壓計內﹐這些數據就一直保存。當您移出電池或電池用盡的情況下﹐儲存的數據將會丟失。
1.
2.
3.
4.
5.

請按壓儲存按鈕。
日期和最新數據將會顯示。
當儲存按鈕被按下,先前的數據會顯示。
當最後一個數據顯示後,顯示屏呈現空白狀態。
壓住儲存按鈕直至“＂標識出現, 即為清除原儲存的數據。

註: 當儲存數據顯示時,按壓 START (開關)按鈕,血壓計便會開始測量。

中文 10

請按壓儲存按鈕。

日期數據什麼是不整脈
UA-789XL血壓計在發生不整脈時也能測量出血壓和脈搏數。所謂不整脈是指測量血壓時﹐脈搏的變化在平均脈搏數 25% 以上。在測量時請安靜保持輕松﹐姿勢坐好﹐不要說話。
)經常在指示器上顯示不停時, 請和醫師商量。

註﹕ 心臟的標識(血壓柱指示器
本血壓柱指示器在測量中顯示壓力的變化。
加壓排除空氣開始膨脹

開始測量
膨脹結束

WHO分類指示器
六個血壓柱指示器分別與下一頁說明的WHO血壓分類一致。

例:

WHO分類指示器
嚴重高血壓

中度高血壓

中度高血壓
輕度高血壓
正常血壓
正常輕度高血壓

最合適: 指示器以現在的數據為依據。顯示與WHO分類一致的區段。

正常血壓值有關血壓的一些知識什麼是血壓？
所謂血壓是血液對動脈血管壁所施加的壓力。當心臟收縮時產生的壓力為收縮壓﹐而當心臟舒張的時候產生的壓力為舒張壓。
血壓以毫米汞柱(mmHg)作為測量單位。當一個人處於安靜狀態和飲食前﹐測量到的血壓代表一
個人的基礎血壓。
中文 11

什麼是高血壓？它是怎樣被控制的？
高血壓是一種動脈異常的高壓力狀態。假若未被及時注意的話﹐會引起許多的健康問題包括休克和心臟病發作。 高血壓可以通過改變生活方式﹐避免緊張和在醫生的指導下服用一些藥物而得到控制。 預防高血壓或者通過下列措施而得到控制。
不要吸煙經常體育鍛煉減少食鹽和脂肪的攝入
定期體檢
維持適當的體重為什麼在家裡測量血壓？
在診所或醫生辦公室測量血壓時容易產生憂慮心理﹐使血壓讀數比在家庭測量的數據高25～30mmHg。在家測量時能減少血壓測量時的一些外部影響因素, 提供給醫生更準確而又完整的血壓紀錄數值。

WHO(世界衛生組織)血壓分類標準WHO(世界衛生組織)設立的評價
高血壓的標準如下圖(不考慮年齡
因素)參考資料：高血壓雜誌1999第17卷第2 期
Ⅲ高血壓(重度)舒Ⅰ高血壓（輕度）壓一個人的血壓可因白天和季節而有些變化。 一般由於白天因素血壓值相差30～50mmHg.。 有高血壓的人, 這種差異可能更加明顯。
通常當工作或運動的時候﹐血壓值會升高。 而在睡眠的時候會下降到最低水平。 所以不要太介意Ⅱ高血壓（中度）

張血壓差異

高於正常
正常
最佳收 縮 壓一次的測量結果。 使用本手冊所說明的方法, 在每天的同一時間測量血壓, 不但知道您自己的血壓值, 而且多次的測量值可以讓您更全面的掌握血壓紀錄資料。 每次測量記錄您的血壓數值時﹐請弄清楚測量的
血時間和日期。 請醫生幫助解釋您的血壓數據。
壓在一天之內的典型動態圖
(每5分鍾測量一次)收縮壓睡眠狀態

舒張壓
下午

中文 12

上午時間

故障檢修
問題

問題原因推薦解決方法

當電源打開時,顯示 電池用盡
屏上沒有任何顯示 電池方法放置不正確袖帶不充氣

更換全部電池放置電池時陰陽極方向要和袖帶上標
的方向一致

電池電壓太低 (低電壓時, 出現本標 更換新電池記。假若電池用盡本標記不會出現。)

本血壓計不能測量, 袖帶扎得不夠緊讀數太高或太低。 當在測量血壓時﹐您移動了手臂或身體。
袖帶位置放置不正確

正確扎緊袖帶
在測量時保持安靜
坐好保持安靜﹐提高您的手保持袖帶和心臟處於同一高度。
假若心臟跳動微弱或不規律性血壓計也很難測到血壓值其它

測到的血壓值和在診所或醫生辦公室測 請參見”為什麼在家裡測量血壓”
量的血壓值不同。
取出電池﹐再正確放進去重新測量。

註: 上述的處置還不能解決問題時, 請與購買的商店聯絡。壇自打開本機或進行修理時就會失去售後
服務的權利, 煩請留意。

維修和保養
請勿打開本血壓計﹐它是採用精密的電子原件和一些複雜的部件製造的。這些原件容易損害。
假若您按照故障檢修還不能排除故障時﹐請向供應商或Ａ＆Ｄ服務中心咨詢。Ａ＆Ｄ將向您提供權威的技術信息﹐備用部件和原件。
本血壓計是為持久耐用而製造的。因此,為使本血壓計能夠發揮最合適的性能和最準確的精度, 請
每兩年接受一次定期維護檢查。維修, 保養請在購買的商店或Ａ＆Ｄ進行。

技術資料型﹕ UA-789XL

測量方法﹕ 示波測量測量範圍﹕ 壓力﹕
脈搏﹕20～280mmHg
40～200 次/分

測量準確性﹕ 壓力﹕
脈搏﹕±3mmHg
±5%

電力供給

﹕ 4×1.5V 電池 (R6P, LR6或AA) 或
交流電轉接器 (TB-242) (未隨附)

測量次數

﹕ 若使用 AA 鹼性電池，室溫 23°C，壓力值 180 mmHg，大約可測量 450 次。

類

中文 13

種

類

﹕ 內部供電ME（醫電）設備 (由電池提供) /Class II (由交流電轉接器提供)
連續操作模式臨床試驗

﹕符合ANSI/AAMI SP-10 1987

EMC﹕IEC 60601-1-2:2007

數據儲存器﹕最近60個的測量值操作條件﹕+10℃至+40℃/ 15% RH 至 85% RH / 800 hPa 至 1060 hPa

運送 /儲存條件﹕ -10℃至+60℃/ 15% RH 至 95% RH

尺寸﹕大約163(W)×62(H)×112(D)毫米

重量﹕大約360克﹐除外電池觸身零件﹕ 袖帶 BF

型

使用壽命﹕ 血壓計﹕袖帶﹕

5 年 (若一天使用六次)
2 年 (若一天使用六次)

交流電轉接器該交流電轉接器用於家庭連接電源。

TB-242

請和當地的銷售商聯繫，購買該交流電轉接器。
必須定期檢查或更換交流電轉接器。

TB-242C

輸入:
輸出:

100-240V
6V
2A

TB-233BF

輸入:
輸出:

240V
6V

2A

附件單獨銷售袖帶目錄編號袖帶尺寸

上臂尺寸

CUF-D-LA
CUF-B-XLA

大型袖帶

32 cm 到 45 cm

超大型袖帶

42 cm 到 60 cm

上臂尺寸:指肩到肘腕之間的上臂外圍尺寸。
交流電轉接器目錄編號插頭

TB-242CTB-242BF

類型C類型BF

註﹕有關技術規格可隨時更改﹐不再提前通知。
EMC表在我們的網站：

http://www.aandd.jp/products/manual/medical/emc_acadapter_ch.pdf

中文 14Asahi 1-243, Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
OX14 1DY United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409