Accurate measurement with sophisticatedautomatic cuﬀ size adjustment mechanism
Easy operation
Small footprint

Application & Point of use
The TM-2657P is the latest model in the range of freestanding fully automatic blood pressure
monitors such as the acclaimed previous models,whose ground-breaking designs have been wellreceived and have created many new applications, places of use, and new markets.
The main reasons for the high reputation of the previous models are their fast and accurate
measurement, compact profile, ease-of-use, and durability. The TM-2657P has been superblystyled in Japan, having a slim and compact shape with increased printout speed, and offers adiversified selection for connectivity. The TM-2657P helps expand market potential, not only for
hospitals, medical clinics and health check-ups, but also for corporate wellness in the workplace,pharmacy services, and fitness centers.

Hospital

Pharmacyservice

Corporatewellness

Medical clinics

Health checkup
system

Accurate measurement at hospitals,
other medical clinics...
In some hospitals, patients can take their own bloodpressure measurement using a fully automatic monitor
before seeing the doctor.
This “self triage" saves the patients' time and lightens
the workload for the medical staff. And with modern
validated BP monitors, the readings are as accurate asif the medical staff took the measurement.

Fitness center

Pick an arm, any arm

Ease of use removes measurement strain

Dual design allows the users to measure witheither arm.

The slim upright design of TM-2657P enables the ideal arm position to makeprecise blood pressure measurements.

Conventional monitors

TM-2657P

Barrier-free design

Small footprint

One button operation

It is convenient for pregnant women,patients with back problems and patientsin wheelchairs.

Slim design allows the users to put it against
a wall or on a desk/counter top without
taking up too much space.

START / STOP button located in themiddle of the unit allows simple
measurement with either hand.

241 mm"Torque Controlled Belt drive Method" (TCBM) adopted in our cuff fastening method

Accurate and reliable oscillometric measurement of bloodpressure on the upper arm relies on correct cuff/bladderdimensions and pressure applied.
The A&D Torque Controlled Belt drive Method (TCBM) holds theupper arm perfectly with the essential personalized cuff
circumference, the same measurement preparation carried out
by clinical nurses.
This optimum ergonomic design enables the users to measure
their blood pressure with accuracy, ease, safety, high speed andcomfort.

Software Development Kit(SDK)

Optional extension communication board(RS-232C, USB, Bluetooth®)

We offer SDK to integrate your application or system
together with our devices• TM-2657-01-EX with MiniDIN 8pin and D-Sub 9pin
• TM-2657-02-EX with USB（TypeA x 1+ TypeB x 1）
• TM-2657-03-EX with D-Sub 9pin• TM-2657-04-EX with D-Sub 9pin and Bluetooth Low Energy

You can request it from here:
https://www.aandd.jp/products/medical/sdk.html

The user can take control of the data and automate the operation.
It can connect to PC, card readers, barcode readers, etc..
The user can also connect TM-2657P with other Bluetooth enabled
devices when using TM-2657-04-EX.
(Please consult with A&D's dealers about the compatibility.)

Reliable thermal printer with open head design
This allows easy paper roll replacement and also helps to avoid possible paper jams.

Various print-out formats

BitmapQR codeTM-2657P provides the users with print-outs (original mark, pictures,messages, etc.) with your ad, pharmacy name, etc..
A large print area is reserved for the advertisement.
The printout can also include measurement information from otherdevices including some personal scales.
(Please consult with A&D's dealers about the compatibility.)

Other printing examples“IHB” (Irregular Heart Beat)

A&D Connect Smart (iOS/ Android)

The symbol for “IHB" is printed when it
detects irregular pulse rate. Body motion
and/or noise can be the reason.
(Note: This is for warning purpose only.
A&D recommends contacting a physician
if you see this symbol () frequently.)

This is a Healthcare App, which can store your vital data to takecontrol of your own health. It scan the QR code on the print as well.

Specifications

Options & Accessories

Model name

TM-2657P

Measurement method

OscillometricAC power supply unit

Switching 100 V~240 V AC 50 Hz~60 Hz

Protection againstelectric shock
Pressure display range

Class I, Type B
0 mmHg –299 mmHg

NIBP measurement range SYS 40 mmHg~270 mmHg, DIA 20 mmHg~200 mmHg, Pulse 30 bpm~240 bpm
Measurement accuracy
NIBP clinical test

Pressure: ±3 mmHg Pulse: ±5 %BS EN1060-4:2004, BHS:1993, +/−5 mmHg SD:8 mmHg

Cuff fastening method

Torque Controlled Belt drive Method (TCBM)

External dimensions

241(W)×324(H)×390(D) mm (9.5(W)×13.0(H)×15.4(D) inches)

WeightApplicable arm
circumference
Cuff size

Approx. 5.5 kg (12.1 lb)

StandAdjustable height stool

Printer paper(5 rolls/set)

Arm cuff cover(5 pieces/set)

Carrying caseRS-232C cable

External input/output
unit RS 2ch

External input/output
unit RS 1ch

External input/output
unit RS+Bluetooth
Low Energy

External input/output
unit USB

REF: TM-ST520

REF: TM-STA001

7 inches (18.0 cm) to 13.8 inches (35.0 cm)
125×300 mm (4.9×11.8 inches)

Display type
Systolic (SYS)
Diastolic(DIA)

3-digit display by LED

Pulse rate(PUL)Counting function
Applied pressure
Pressure detection method

Display of number of measurements to date: up to 999,999 times
Automatic inflation by air pump
Capacitance type pressure transducer

Operating conditions

Temperature: +10 °C~+40 °C humidity: 30 %RH~85 %RH, no condensing

Transport/Storageconditions

Temperature: –20 °C~+60 °C humidity: 10 %RH~95 %RH, no condensing

Safety device(Electrical)

Quick release when START/STOP button is pressed
Quick release when FAST STOP button is pressed

Clock function

Clock Display, Date & Time (2015 ~ 2050)

Dimensions (mm)

REF: AX-PP147-S

REF: TM2657-10

REF: TM-2657-01-EX

REF: TM-2657-04-EX

REF: AX-134005759-S

REF: AX-KO1371-200

REF: TM-2657-03-EX

REF: TM-2657-02-EX

A&D Company Limited.(JAPAN)

A&D AUSTRALASIA PTY LTD(Australia)

A&D ENGINEERING, INC.(USA)A&D RUS CO., LTD.(Russia)

A&D Instruments Canada Inc.(CANADA)

A&D Technology Trading(Shanghai)Co., Ltd(China)

A&D INSTRUMENTS LIMITED(United Kingdom)

A&D INSTRUMENTS INDIA PRIVATE LIMITED(India)

URL:aandd.jp

URL:andonline.com

URL:autocontrol.com

URL:andprecision.com

URL:andaustralasia.com.au
URL:and-rus.ru

URL:aanddtech.cn

URL:aanddindia.in*TM2657P-ADCC-07-PR06-245001GPREF*ADC-TM2657P