表4

表面（山折り面）

NEW

センター折り表1（表紙）

Weighing Indicators

AD-4405A/4406A/4407A
New and improved!

Easy to build a Weighing System
to your own specifications using
the featured Weighing Indicators
AD-4405A

AD-4407A

AD-4406A表2

中面（谷折り面） センター折り

A multitude of Weighing System applications can beachieved with A&D's featured Weighing Indicators

AD-4405A
Built-in Printer as optional add-on foryour weighing data recording●Bright green VFD 20mm (height) display●Optional dot matrix compact printer●10 keys for preset tare andunit weight setting, etc.
●Counting function●Standard RS-232C

AD-4406A

Rear panel
With optional built-in printerExpand your weighing applications
by Battery operation●High cost-effectiveness●Big LCD 25mm (height) display●Multi power supply
Alkaline C type battery:
Battery life 180 hours (LC 350Ω ✕ 1)
Battery life 70 hours (LC 350Ω ✕ 4)
Optional AC adapter●F1, F2 selectable function
keys●Optional RS-232C●Load cell 350Ω ✕ 4●Optional stand available

Rear panel

With optional standAD-4407A
Stainless Steel Enclosure to meet IP65
Dust & Waterproof standards forweighing applications indemanding environments●Bright green VFD
20mm (height) display●Bracket standard●10 keys for preset tare andunit weight setting, etc.
●Counting function●Standard RS-232C

Rear panel

With standard stand

表3

表2

中面（谷折り面）

表3

センター折りAD-4405A/4406A/4407A

A multitude of Weighing System applications can beachieved with A&D's featured Weighing Indicators

Weighing Indicators

Common Features
 Universal Flexi Coms
UFC allows you to customize the protocol of the printing format
as well as outputting data with RS-232C
 Multi interval
2 weighing ranges selectable Accumulation function
Accumulative data is stored into memory automatically or manually Comparator function Upper/Lower, HiHi/Hi/OK/Lo/LoLo & comparison
Comparator output to optional relays
 Simple batch weighing
Automatic batching and customer programming capability
 Buzzer function
 Code memory function

APPLICATIONS with LOAD CELL
You can use our load cells as a load sensor to build your own desktop, bench, floor or truck scales
as per your preferences.

AD-4405A/4406A/4407A
Weighing Indicators+LOAD CELL
LCB05
LC4221
LCC07
LCC11
LCC12
LCM13LCS15

Waterproof
Weighing
SystemWeighingPlatform
SB Series

LCB03
LCB04
LC4101
LC4102
LC4103

Bench Scale
Platform Scale

LC4204
LC4212
LCB04

Floor ScaleAPPLICATIONSLC2224LCC07
LCC11
LCC12Truck Scale
Tank Scale表4AD-4405A/4406A/4407A
Weighing Indicators

Specifications
AD-4405A

AD-4406A
AD-4407A
0.15µV/d—35mV to +35mV
DC 5V±5% 120mA, remote sensing
DC 5V±5% 60mA, remote sensing(Up to 8 load cells at 350Ω/load cell)(Up to 4 load cells at 350Ω/load cell)±0.02µV/ºC typ. ±0.1µV/ºC max.
Zero Temperature Coefficient±3ppm/ºC typ. ±15ppm/ºC max.
Span Temperature Coefficient
0.005% of full scale
Non-Linearity
0.15µVp-p typ.
Input Noise
10MΩ or more
Input Impedance
Delta Sigma
A/D Conversion Method
16,000,000 counts
A/D Resolution
20,000 (permissible 40,000)
Maximum Display Resolution
10 times/sec.
A/D Conversion Rate
7-segment, 25mm(H) LCD
Main Display
7-segment, 20mm(H) VFD
7-segment, 20mm(H) VFD
7 to 10V DC
Power
AC 100/120/200/230V
AC 100/120/200/230V
AC adaptor or—15~+10% (50/60Hz)—15~+10% (50/60Hz)
C battery ✕ 6Less than 10VA
Consumption
20VA
20VA
180H (1 load cell)
Battery Life__70H (4 load cells)(Alkaline Battery)—10 to +40ºC (14ºF to 104ºF)
Operating Temperature
Less than 85% RH (non-condensing)Operating Humidity
275(W) ✕ 111(H) ✕ 159(D) mm
199(W) ✕ 206(H) ✕ 80(D) mm
192(W) ✕ 96(H) ✕ 87(D) mm
Physical Dimensions
10.83(W) ✕ 4.37(H) ✕ 6.26(D) inches 7.56(W) ✕ 3.78(H) ✕ 3.43(D) inches 7.83(W) ✕ 8.11(H) ✕ 3.15(D) inches
Approximately 1300g / 2.87 lb
Approximately 2700g / 5.95 lb
Approximately 650g / 1.43 lb
Net Weight
Input Sensitivity
Input Signal Range
Load Cell Excitation

Options

Accessories

AD-4405-03
AD-4405-05
AD-4405-06
AD-4405-07
AD-4405-08

RS-422/485 and 3 Relay Outputs
RS-232C, 3 Relay Outputs and 3 Control Inputs
Built-in PrinterAnalog Output (4-20mA)
RS-232C, Current Loop Output,
3 Relay Outputs and 1 Control Input

AD-4406-03
AD-4406-04
AD-4406-05
AD-4406-07
AD-4406-08

RS-422/485 and 3 Relay Outputs
RS-232C
RS-232C, 3 Relay Outputs and 3 Control Inputs
Analog Output (4-20mA)
RS-232C, Current Loop Output,
3 Relay Outputs and 1 Control Input
Display Stand

AD-4406-11
AD-4407-03
AD-4407-05
AD-4407-07
AD-4407-08

AD-8118C
AD-8121B
AD-4379SUS
AD-4380
AD-4382

Universal PrinterDot Matrix Compact PrinterSumming Box (4 points)
Junction Box
Summing Box (2 points)

RS-422/485 and 3 Relay Outputs
RS-232C, 3 Relay Outputs and 3 Control Inputs
Analog Output (4-20mA)
RS-232C, Current Loop Output,
3 Relay Outputs and 1 Control Input

Appearance and/or specifications subject to change for improvement without notice.

…Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku, Tokyo 170-0013 JAPAN
Telephone:［81］(3) 5391-6132 Fax:［81］(3) 5391-6148
http://www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:(408) 263-5333 Fax:
［1］
［1］(408) 263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:［61］(8) 8301-8100 Fax:［61］(8) 8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOMTelephone:［44］(1235) 550420 Fax:［44］(1235) 550485〈German Sales Office〉

Hamburger Straße 30 D-22926 Ahrensburg, GERMANY
Telephone:［49］(0) 4102 459230 Fax:［49］(0) 4102 459231

A&D Korea Limited

A&D RUS Co., Ltd.

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone:(495) 937-33-44 Fax:
［7 ］
［7］(495) 937-55-66

A&D Instruments India Private Limited

509 Udyog Vihar Phase V Gurgaon-122 016,Haryana, INDIA
Telephone:［91］(124) 471-5555 Fax:［91］(124) 471-5599

817, Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil,Yeongdeungpo-gu, Seoul, 150-749, KOREA
Telephone:［82］(2) 780-4101 Fax:［82］(2) 782-4280✽ AD440567A-ADCC-01-ZW1-15103