Multi-function Infrared Thermometer
Model UT-801

Instruction Manual ENGLISH
Original

Manuel d’instructions FRANÇAIS
Traduction

Manual de Instrucciones
TraducciónESPAÑOL

Manuale di Istruzioni ITALIANO
Traduzione

使用手冊 中文翻譯

1WMPD4003399B‫ﻋرﺑﻲ دﻟﯾل اﻟﺗﻌﻠﯾﻣﺎت‬‫ﺗرﺟﻣﺔ‬Contents
Dear Customers .............................................................................................................................................................................................2
Preliminary Remarks......................................................................................................................................................................................2
Precautions .....................................................................................................................................................................................................2
Symbols ..........................................................................................................................................................................................................3
Parts Identification ..........................................................................................................................................................................................4
Functions.........................................................................................................................................................................................................5
Installing/changing batteries ..........................................................................................................................................................................7
Measurements................................................................................................................................................................................................8
Troubleshooting............................................................................................................................................................................................11
Care & Maintenance ....................................................................................................................................................................................12
Technical Data..............................................................................................................................................................................................13
Appendix: EMC Information ........................................................................................................................................................................14

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D thermometer, one of the most advanced thermometers available today. Designed
for ease of use and accuracy, this device will facilitate your health care regimen.
We recommend that you read through this manual carefully before using the device for the first time.

Preliminary Remarks▢ This device conforms to the European Directive 93/42 EEC for Medical Products. This is proved by the CE conformity marking(1639: The identification number of the Notified Body).
▢ Environment for use: This device is designed for use indoors.
▢ This device (the infrared ear/forehead thermometer) can measure the infrared heat generated by eardrum/temple area and surrounding
tissues to reflect patient’s body temperature accurately.

Precautions▢ Precision components are used in the construction of this device. Extremes in temperature, humidity, direct sunlight, water, shock or dust
should be avoided.
▢ There is no gender and age limitation for using the infrared thermometer.
▢ Do not let children use the device by themselves and do not leave the device in a place within the reach of infants.
▢ Intended operator: At least 11 years old (5 years intensive reading experience), no maximum.
English 2

▢ This thermometer has been designed for everyday use. It’s not meant to replace a visit to the doctor. Please also remember to compare
the measurement result to your regular body temperature. Please consult with doctor if you have health concerns.
▢ Wireless communication devices, such as home networking devices, mobile phones, cordless phones and their base stations, and walkietalkies can affect this thermometer. Therefore, a minimum distance of 3.3 meters (11 feet) should be kept from such devices.
▢ Measurements may be distorted if the device is used close to televisions, microwave ovens, X-ray or other devices with strong electricalfields.
▢ There are small parts that may be a choking hazard if swallowed accidentally by infants.
▢ No AP/APG (not suitable for use in the presence of flammable anesthetics or oxygen)

Symbols

IP22

Please read the instructions for use

EC directive medical device label

Type BF: Device

WEEE label

Classification for water ingress and particulate matter

Manufacturer

EU-representative

CautionStand-byBattery Recycling

English 3

Paper Recycling

Parts Identification
When first time using, please pull out
the isolation sheet.
Last memory icon

Last memory

Surface icon

Cap

SCAN buttonStand-by(ON/MEM button)

Forehead icon

Ear icon

Serial number :SN Byymmxxxxx

Display

Battery icon

Memory icon

B: fixyy: year
mm: months
xxxx :production
number

Unit icon

Mute icon

Isolation sheet
English 4

Battery coverFunctions
Ear temperatureThis thermometer has been designed for everyday use. It’s not meant to replace a visit to the doctor. Please also remember to compare
the measurement results to your regular body temperature.
Forehead TemperatureThis thermometer has been designed for everyday use. It’s not meant to replace a visit to the doctor. Please also remember to compare
the measurement results to your regular body temperature. Please consult with doctor if you have health concerns.
Room temperature
Suitable ambient temperature is important for the baby and patient. The thermometer always helps you recognize the room temperature.
Surface Mode
The surface mode shows the actual and undusted surface temperature which is different from the body temperature. It can help you
monitor if the object temperature is suitable for the baby or patient, for example the baby’s milk.
Fever Indication
If the thermometer detects a body temperature above 37.5°C, there will be a long beep followed by three short beeps to warn the user ofpotential fever.
Last Reading
When you get a new temperature reading in ear mode or forehead mode, the last reading will be shown on the screen (in the top rightcorner) with the last reading icon.

English 5

Memory Locations
There are a total of 25 memory sets for ear measurements. --- When powered on, press the “ON/MEM” button to see the temperatures
stored with memory icon.
°C / °F Switch
In “Power Off” mode, press and hold the “SCAN” button, then press the ON/MEM button for 3 seconds. The temperature icon “°C” willswitch to “°F”. You can also use the same process to change from °F to °C.
Mute mode
The buzzer setting defaults to on. You can toggle the buzzer on/off in mute mode. With the power on, press and hold the “ON/MEM” button
for 3 seconds. The mute icon will flash on the display. Release the “ON/MEM” button to turn on MUTE. Now you will not hear beeps. You
can also use the same process to turn off the mute function.
NOTE: If you keep pressing the "ON/MEM" button for 5 seconds after the mute icon flashes, the device will turn off WITHOUT setting mute.
Auto power off
Device will automatically shut off(stand-by) if left idle for more than 1 minute. And the device can only be powered off automatically.
Device is always displayed the ambient temperature in stand-by and you cannot turn off all displays on LCD.

English 6

Installing/changing batteries
1. Open the battery cover: Make sure the thermometer is in ”forehead mode”. Insert a coin into the twist cover at the rear of the device andtwist it clockwise. The battery is located under the cover.
2. Hold the device and flip the battery out with a small screwdriver.
3. Insert the new battery under the metal hook on the left side and press the right side of the battery down until you hear a click.
4. Replace the battery cover.

CAUTION▢ Keep the battery away from children.(This device is supplied with one CR2032 lithium cell.)▢ Keep the positive (+) side up and the negative (-) side pointed down.
▢ Comply with local regulations for the disposal of used batteries.

English 7

Measurement
1. Always make sure the probe is clean and undamaged.
2. Power on: Press the “ON/MEM” button.
3. Mode selection:● Ear mode:

Press the “ON/MEM” button, the default mode of thermometer is ear mode. The thermometer is can be used after
you see the ear icon on the display and hear two beep sounds. In this mode, you can measure the body temperatureby ear measurement.

● Forehead mode: After power on, press and hold the “ON/MEM” button, and press the “SCAN” button one time for forehead mode. The
forehead icon will display and you will hear two beeps. In this mode, you can measure the body temperature byforehead measurement.
● Surface mode:

After power on, press and hold the “ON/MEM” button, and press the “SCAN” button one time for surface mode. The
surface icon will display. In this mode, you can measure surface temperature of an object.

4. Measuring temperature:
Ear temperature measuring. Points for attention:(1) Gently pull the ear back to straighten the ear canal and snugly position the probe into the ear canal, aiming
towards the membrane of the eardrum to obtain an accurate reading.
(2) Press and hold the “SCAN” button until you hear a beep sound. Remove the probe from the ear and readthe temperature measurement on the LCD.
English 8

Pull

Scan

NOTES▢ It is recommended that you measure 3 times with the same ear. If the 3 measurements are different, select the highest temperature.
▢ To avoid the risk of cross contamination, please clean the probe according to the “Care & Maintenance” section after each use.
▢ Temperature varies in healthy persons and different parts of the body can have differences between 0.2 to 1°C.
Forehead temperature measuring: Points for attention:(1) The temporal artery, a major artery of the head, is connected to the heart via the carotid artery. This
thermometer is designed to measure the skin surface around the temporal artery.
(2) Place the thermometer on the temple (see picture). (You can choose left or right temple.)(3) Press the “SCAN” button, and gently scan around the temple area. While scanning, you will hear a scansound, which indicates the newest measurement is taking place. Measurement is complete when theforehead icon stops flashing after a short beep. Measurement time may be between 5 ~ 8 seconds and up
to 30 seconds) depending on how much time the device needs to get the correct forehead temperature.
NOTES▢ Forehead temperature is displayed in oral mode. This mode converts the forehead temperature to display its “oral-equivalent” value.
▢ Before measurement, please stay in a stable environment for 5 minutes and avoid exercise and bathing for 30 minutes.
▢ Remember to keep the temple area clean and free from sweat and cosmetics and avoid scars while taking the temperature.

English 9

Measuring temperature in surface mode:(1) When you press the “SCAN” button, you will get the real time temperature immediately. If you press and hold the “SCAN” button, themeasurement will be continuously updated.
(2) Applications include temperature measurements of water, milk, cloth, skin or other objects.
NOTE
This mode shows actual, unadjusted surface temperatures, which are different from body temperature.
5. After measurement:(1) Power off:
Device will automatically shut off(stand-by) if left idle for more than 1 minute. And the device can only be powered off automatically.
(2)Clean the probe after each use to ensure accurate readings and avoid cross contamination. (See the section on "Care & Maintenance"for details.)(3) Device is always displayed the ambient temperature in stand-by and you cannot turn off all displays on LCD.
NOTES▢ This thermometer has been designed for everyday use. It’s not meant to replace a visit to the doctor. Please also remember to compare
the measurement result to your regular body temperature. Please consult with doctor if you have health concerns.
▢ Holding the thermometer too long may cause a higher ambient temperature reading of the probe. This could make the bodytemperature measurement lower than usual.

English 10

Troubleshooting
Error

Problem

Recommended Action
Take out the battery, wait for 1 minute andreinsert it. If this message reappears,
contact the retailer for service.
Wait until all the icons stops flashing before
measuring.

Error 5 ~ 9, the system is not functioning properly.
Measurement before device stabilization.

Allow the thermometer to rest in a room forat least 15 minutes at room temperature:
10°C to 40°C (50°F to 104°F).

The ambient temperature is >40°C (104°F)
The ambient temperature is <10°C (50°F)
In ear/forehead mode: Temperature taken is higher than +42.2°C (108°F)
In surface mode:
Temperature taken is higher than +80°C (176°F)
In ear/forehead mode: Temperature taken is lower than +34°C (93.2°F)
In surface mode:
Temperature taken is lower than –22°C (–7.6°F)

Please select the target within
specifications. If a malfunction still occurs,
please contact the nearest retailer.

Device cannot be powered on to the ready stage.

Replace battery with a new one.

CAUTION
Do not open this device. If you cannot fix the problem using the troubleshooting instructions, contact the dealer or customer service.
English 11

Care & Maintenance▢ After measurement, please use a cotton swab with alcohol (70% concentration) to clean the lens (on the inside of the probe).
▢ Allow the probe to fully dry for at least several minutes. Then reattach the cap.
▢ This device should be stored at a temperature between -20 to +50°C and humidity 85% or less.
▢ Keep this device dry and away from any liquids and direct sunlight.
▢ The probe should not be submerged into liquids.
NOTES▢ The probe is the most delicate part of the thermometer. Use with care when cleaning the lens to avoid damage.
▢ Please check the device for damage if it is dropped. If you are unsure how to, please send the complete device to the nearest retailer forrecalibration.
▢ Holding the thermometer too long may cause a higher ambient temperature reading of the probe. This could make body temperature
measurement lower than usual.

English 12

Technical Data
Dimensions

125 mm × 37 mm× 22 mm

WeightApprox. 41g, excluding the battery.

Temperature measurement range Ear/Forehead mode: 34 to 42.2°C(93.2 to 108°F), Surface mode: -22 to +80°C(-7.6 to 176°F),
Operating temperature range

10 to 40°C (50 to 104°F) ,15 to 85%RH

Storage temperature range

Device should be stored at a room temperature between -20 to +50°C and humidity 85% or less.
Transportation temp. shall be less than 70°C ,RH≦95%, Atmospheric pressure:800 to 1013hPa.

Accuracy

Ear/Forehead mode: ±0.2°C (0.4°F) within 35 to 42°C (95 to 107.6°F) (Ambient Temp: 15 to 35°C),±0.3°C (0.5°F) for other range.
Surface mode: ±0.3°C (0.5°F) within 22 to 42.2°C (71.6 to 108°F)
others ±4% or 2°C (4°F) whichever is greater.

Memory25 memory slots and last memory.

Battery

CR2032×1

・Battery life: around 3000 continuous readings. ・Expected Service Life: 4 years.
・This thermometer converts the forehead/ear temperature and displays the “oral equivalent”.
・Comply with ASTM E1965-98, EN ISO 80601-2-56, IEC/EN60601-1-2(EMC), IEC/EN60601-1(Safety) standards, ISO10993, RoHS.
NOTE: Specifications are subject to change for improvement without prior notice.

English 13

Appendix: EMC Information
Manufacturer’s declaration-electromagnetic emissions
The UT-801 is intended for use in the electromagnetic environment (for home healthcare) specified
below. The customer or the user of the UT-801 should assure that it is used in such an environment.

Emission test

ComplianceElectromagnetic environment-guidance(for home healthcare environment)

RF emissions
CISPR 11

Group 1The UT-801 uses RF energy only for its internal function. Therefore, i t s
R F emissions are very low and are not likely to cause any interference innearby electronic equipment.

RF emissions
CISPR 11Class BThe UT-801 is suitable for use in all establishments, including domestic
establishments and those directly connected to the public low-voltage power
supply network that supplies buildings used for domestic purposes.

English 14

Manufacturer’s declaration-electromagnetic immunity
The UT-801 is intended for use in the electromagnetic environment (for home healthcare) specified below.
The customer or the user of the UT-801 should assure that it is used in such an environment.

Immunity test

Electrostatic
discharge(ESD)
IEC61000-4-2
Power frequency(50,60 Hz)
magnetic field
IEC 61000-4-8

IEC 60601
test level

Compliance level

Contact:±8 kV

Contact:±8 kV

Air±2 kV,±4 kV,

Air±2 kV,±4 kV,±8 kV,±15 kV±8 kV,±15 kV

30 A/m50 Hz or 60 Hz

30 A/m50 Hz and 60 Hz

English 15

Electromagnetic
environment-guidance(for home healthcare environment)
Floors should be wood, concrete or
ceramic tile. If floors are covered withsynthetic material, the relative humidity
should be at least 30%
The UT-801 power frequency magnetic
fields should be at levels characteristicof a typical location in a typical homehealthcare environment.

Manufacturer’s declaration-electromagnetic immunity
The UT-801 is intended for use in the electromagnetic environment (for home healthcare) specified below.
The customer or the user of the UT-801 should assure that is used in such and environment.
Immunity test

IEC 60601
test level

Electromagnetic environment-guidance(for home healthcare environment)

Compliance level

Recommended separation distance:d = 1,2 √P
d = 1,2 √P
Radiated RF
IEC 61000-4-3

10 V/m

10 V/m80 MHz – 2,7 GHz

80 MHz – 2,7 GHz

80 % AM at 1 kHz

80 % AM at 1 kHz

80MHz to 800 MHz

d = 2,3 √P 800MHz to 2,7 GHz

Where P is the maximum output power rating of the transmitter inwatts (W) according to the transmitter manufacturer and d is therecommended separation distance in metres (m).
Field strengths from fixed RF transmitters, as determined by anelectromagnetic site survey, a should be less than the compliancelevel in each frequency range.b
Interference may occur in the vicinity of equipment marked
with the following symbol:

NOTE1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures,
objects and people.
a Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio,
AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment
due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the
UT-801 is used exceeds the applicable RF compliance level above, the UT-801 should be observed to verify normal operation. If abnormal
performance is observed, additional measures may be necessary, such as re-orienting or relocating the UT-801.

English 16

Recommended separation distance between

portable and mobile RF communications equipment and the UT-801
The UT-801 is intended for use in an electromagnetic environment (for home healthcare) in which radiated RF disturbances are controlled.
The customer or the user of the UT-801 can help prevent electromagnetic interference by maintaining a minimum distance between portable
and mobile RF communications equipment (transmitters) and the UT-801 as recommended below, according to the maximum output powerof the communications equipment.
Rated maximum output powerof transmitter

Separation distance according to frequency of transmitterm
150 kHz to 80 MHz

80 MHz to 800 MHz

800 MHz to 2,7 GHz

d =1,2√P

d =1,2√P

d =2,3√P

0,01

N/A

0,12

0,23

0,1

N/A

0,38

0,73

W

1

N/A

1,2

2,3

10

N/A

3,8

7,3

100

N/A

12

23

For transmitters rated at a maximum output power not listed above, the recommended separation distance d in meters (m) can be estimatedusing the equation applicable to the frequency of the transmitter, where p is the maximum output power rating of the transmitter in watts (W)according to the transmitter manufacturer.
NOTE1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures,
objects and people.

English 17

Manufacturer's declaration-electromagnetic immunity
Test specifications for ENCLOSURE PORT IMMUNITY to RF wireless communications equipment

The UT-801 is intended for use in the electromagnetic environment (for home healthcare) specified
below. The customer or the user of the UT-801 should assure that it is used in such an environment.
Maximuma)Test frequency

Band

(MHz)

(MHz)

385

380 –390450

430 – 470

Servicea)

TETRA 400

GMRS 460,
FRS 460

Distance(m)

IMMUNITY

Compliance LEVELTEST LEVEL

(V/m)

(V/m)

(for home healthcare)

Pulse
modulation b)
18 Hz

1,8

0,3

27

27

FM c)
5 kHz deviation
1 kHz sine

2

0,3

28

28

LTE Band
13,17

0,2

0,3

9

9

2

0,3

28

28

modulation b)
217 Hz

GSM 800/900,

810

930

power

Pulse
704 – 787

780

870

b)

(W)

710
745

Modulation

800 – 960

TETRA 800,

Pulse

iDEN 820,

modulation b)

CDMA 850,

18 Hz

LTE Band 5

English 18

GSM 1800;

1 720
1 845

CDMA1900;
1 700 –
1 990

1 970

GSM1900;

Pulse

DECT;

modulation b)

LTE Band
1,3,4,25;

217 Hz

2

0,3

28

28

2

0,3

28

28

0,2

0,3

9

9

UMTS
Bluetooth,

2 450

2 400 –
2 570

WLAN,

Pulse

802.11 b/g/n,

modulation b)

RFID 2450,

217 HzLTE Band 7
5 240
5 500
5 785

5 100 –
5 800

WLAN 802.11
a/n

Pulse
modulation b)
217 HzNOTE
If necessary to achieve the IMMUNITY TEST LEVEL, the distance between the transmitting antenna and the ME
EQUIPMENT or ME SYSTEM may be reduced to 1 m. The 1 m test distance is permitted by IEC 61000-4-3.
a)
b)
c)

For some services, only the uplink frequencies are included.
The carrier shall be modulated using a 50 % duty cycle square wave signal.
As an alternative to FM modulation, 50 % pulse modulation at 18 Hz may be used because while it does not represent
actual modulation, it would be worst case.

English 19

MEMO

A&D INSTRUMENTS LIMITED

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409

A&D Instruments India Private Limited

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel：91(124)471-5555 Fax：91(124)471-5599

Radiant Innovation Inc. Http://www.radiantek.com.tw

1F, No.3, Industrial E. 9th Rd., Science-Based Industrial Park,
HsinChu 300, Taiwan