Product Datasheet

Arium®
Advance EDI
Ultimate Reliable
Electrodeionisation---Advantages
Reliable – Consistently high Type 2 water quality thanks tothe latest EDI technology
Time-saving – Use of innovative Bag technology, eliminatescostly tank cleaning
Optimized water consumption – Automatic with iJust
Easy to use – Display with touch function and intuitive
menu

Product Description
The Arium® Advance EDI provides Type 2 water inconsistently high quality. By using the integrated, innovative
iJust function, it does not only optimizes water consumption
but also ensures that natural resources are usedconscientiously. In contrast to conventional water purification
systems, the unique touch display with intuitive menu
navigation makes it extremely easy to use.
With a flow rate of 5 or 10 L/h, the automatic RO membrane
backflush, the latest EDI technology and a constant flow rate,
the Arium® Advance EDI is the ideal choice for general anddemanding laboratory applications.

Consistently High Type 2 Water Quality

iJust

In addition to prefiltration and purification by reverse
osmosis modules, an electrochemical deionization is carriedout as a third purification step. By means of this modern EDItechnology, the Arium® Advance EDI reliably guarantees ahigh effective removal of ions contained in the feed water.

iJust stands for innovative technology that optimizes
water production. Amongst other things, the intelligent
Arium® software controls a valve on the concentrateoutlet in accordance with the data that have been entered
for CaCO3 and CO2. iJust therefore optimizes the productwater qua­lity and water consumption.

Innovative Bag Technology
The pure water is stored in the enclosed Arium® Bagtank
system. This guarantees optimal storage of the pure waterand protects against secondary contamination. Timeconsuming tank cleaning intervals are eliminated thanks
to the interchangeable bag.

---High product water quality at all times
Optimized, economical water consumption
Guarantees a longer life of the downstream ultrapure
water systems

Display With Touch Function
Simply navigate intuitively in the easy-to-use and clear
menu by lightly touching the display – even with gloves.

Technical Specifications
Dimensions [W × H × D]

35.0 × 50.1 × 45.1 cm

Empty weight

19 kg

Operating weight

26 kg

Power supply

100 – 240 VAC (±10%); 50 – 60 Hz, 130 VA (max.)

Operating temperature

2 °C – 35 °C at max. 80% relative humidity

Storage temperature

5 °C – 45 °C at max. 80% relative humidity

Feed Water Quality
Exclusively potable tap water pursuant to the drinking water standards of the USA, the European Union, or Japan.
Input pressure1

2.0 – 6.9 bar

Temperature2 – 30 °C

Specific conductivity< 1,500 µS/cm compensated to 25 °C

TOC< 2,000 ppb

Max. total hardness (max. CaCO3)

360 ppm

Free chlorine

4 ppm

Iron (total Fe content)< 0.1 ppm

Manganese< 0.05 ppmAluminum< 0.05 ppm

CO2 in solution≤ 40 ppm

Fouling Index (SDI)<5Turbidity< 1 NTU

pH value4 – 10

1 Dynamic pressure | flow pressure 100 L/h2

EDI module
RO module #1
Conductivity measurement (productwater | permeate) (RO)Conductivity
measurement(feedwater)

Conductivity
measurement(EDI)*Softener

RO module #2

Inlet(feed water)
Outlet(waste water)
RO pretreatment
cartridge

PumpArium®
Bagtank

Flowchart Arium® Advance EDI (H2O-EDI-2-T)

Water Applications
Advance EDI
Water Quality
Type 2 Pure Water

Lab Water System by Daily Water Consumption
Type 2 pure water approx. 120 Liter/day (5 L/h)
Type 2 pure water approx. 150 Liter/day (10 L/h)Feed Application
Feed ultrapure water systems
Feed distilled systems
Water for Laboratory devices (Autoclaves | Washing Machine etc.)

General Laboratory Application
Buffer, media and pH solutions
Histology
ELISA (Enzyme-Linked Immunosorbent Assay)
AAS (Atomic Absorption Spectroscopy)
Solutions for chemical analysis and synthesis

-----

3

Product Water Quality
Water purification method

Particle filtration, adsorption using spherical activated carbon, catalyst, reverse osmosis,
softening electrochemical deionization

Water typeType 2 pure water

Production output²5 L/h or 10 L/hWater dispensing flow rate⁴

Up to 3 L/min

Typical conductivity¹0.2 – 0.07 µS/cm compensated to 25 °C

Typical resistivity¹5 – 15 MΩ × cm compensated to 25 °C

Retention of dissolved organic
substances (MW > 300 Dalton)

99%

Particle and microorganism retention > 99%
1 Depending on the feed water quality (CO2 ≤ 40 ppm) and temperature²D epending on the feed water pressure, temperature, and condition of the RO module(s)³ Depends on the type of organic contamination in the feed water
⁴ When using an Arium® Bagtank design with pump, depending on hydrostatic pressure, connected accessories or end filter

Additional product water specifications when connected to an Arium® Smart Station¹ with final filter
Particle content²

No particles > 0.22 µm

Bacteria²< 0.001 CFU/mL

Endotoxins³< 0.001 EU/mL

RNase concentration³< 1 pg/mL

DNase concentration³< 5 pg/mL

Water dispensing flow rate⁴

Up to 2 L/min

Volume-controlled dispensing

0.05 – 50 L in 50 mL steps

1 Connected to an Arium® Bagtank² When using an Arium® Sterile Plus (Sartopore® 2 150 final filter)³ When using an Arium® Cell Plus final filter
⁴ Depending on the connected Arium® Bagtank, hydrostatic pressure, connected accessories and end filter.

Accessories and final filters for the Arium® Smart Station can be found in the Arium® Smart Station data sheet

Ordering Information
Arium® Advance EDI for the production of Type 2 pure water
Scope of supply: 1 Arium® Advance EDI, RO (reverse osmosis) module(s), connection kit and water guard
Order numberDescription

H2O-EDI-1-T

Arium® Advance EDI benchtop device, flow capacity 5 L/h

H2O-EDI-1-B

Arium® Advance EDI wall-mounted device, flow capacity 5 L/h

H2O-EDI-2-T

Arium® Advance EDI benchtop device, flow capacity 10 L/h

H2O-EDI-2-B

Arium® Advance EDI wall-mounted device, flow capacity 10 L/h

4

Accessories
Arium® Bagtank---

Innovative tank system
Integrated ventilation filter with non-return valve provides
reliable protection against CO₂ pollution
High flexibility through the 4 rollers available as an option
Easy and fast exchange of the Arium® Bags
No chemical cleaning required

Description
The pure water is stored in the innovative enclosed Arium®
Bagtank system. This system protects the prepared purewater against secondary contamination. The Sartorius
Bagtank system enables consistent water quality over a
prolonged period, thereby ensuring permanent,
reproducible results. Unlike conventional water reservoirs,
the Arium® Bag eliminates time consuming tank cleaningintervals thanks to the interchangeable technology.
Arium® Bagtanks are housings which are equipped withArium® Bags. The Arium® Bagtanks are available in 20 L,
50 L, and 100 L volumes. Their design is adaptable andsaves space in any laboratory environment, and theoptional rollers make this an extremely flexible system.
Integrated distributor pumps are a standard component of
the 50 L and 100 L Bagtanks. A distributor pump is alsoavailable as an option for the 20 L Bagtank. In addition, awall holder for the space-saving and user-friendly
installation of this tank is also available.

Water dispensing flow rate
With pump¹

up to 3.0 L/min

With pump, remote dispenser andSterile Plus filter¹

up to 2.0 L/min

Without pump²

up to 1.5 L/min

Pump pressure

3 bar1 Bagtank 20 is supplied without a pump as standard, pump optionally available
2 Value only applies to Bagtank 20, dispensing site at the same height or lower than the tank outlet-Intended Use
Device type:
Arium® Advance EDI

5

Technical Specifications |Ordering Information
Order number

Description

H2O-AOV-203

Arium® Bagtank 20 L, without pump,1 pc

H2O-AOV-503

Arium® Bagtank 50 L, with pump
240 VAC, 50 Hz, 1 pc

H2O-AOV-50-US3

Arium® Bagtank 50 L, with pump
115 VAC, 60 Hz, 1 pc

H2O-AOV-50-W3

Arium® Bagtank 50 L, without pump,1 pc

H2O-AOV-1003

Arium® Bagtank 100 L, with pump
240 VAC, 50 Hz, 1 pc

H2O-AOV-100-US3Arium® Bagtank 100 L, with pump
115 VAC, 60 Hz, 1 pc

H2O-AOV-100-W3Arium® Bagtank 100 L, without pump,1 pc

Empty weight without Arium® Bag |Operating weight with filled Arium® Bag

H2O-ADP-20

Pump Arium® Bagtank 20 L, 240 VAC,50 Hz, 1 pc

Bagtank 20

19 kg | 40 kg

H2O-ADP-20-US

Bagtank 50

33 kg | 84 kg

Pump Arium® Bagtank 20 L, 115 VAC,60 Hz, 1 pc

Bagtank 10047 kg | 148 kg

H2O-ATR

Rollers for Arium® Bagtank 50 & Bagtank 100, including fastening material,
4 pcs

H2O-CBS-20

Arium® 20 L Bag for Arium® 20 L
Bagtank, 2 pcs

H2O-CBS-50

Arium® 50 L Bag for Arium® 50 L and100 L Bagtank, 2 pcs

Materials
Bagtank

Stainless steel | plastic

Bag

S71 film

TubingPE | silicone

Dimensions, excluding rollers and wall bracket [H × W × D]
Bagtank 20

80.8 × 16.6 × 43.7 cm

Bagtank 50

85.2 × 25.4 × 58.7 cm

Bagtank 100

85.2 × 51.4 × 58.7 cm

Bag 20 L

86.5 × 43.0 cm

Bag 50 L

90.0 × 58.1 cm

Number of bags per tank
Bagtank 20

1 × 20 L

Bagtank 50

1 × 50 L

Bagtank 100

2 × 50 L

Power supply1

240 VAC (±10%), 50 Hz, 120 VA (max.)

Power supply
US versions1

115 VAC (±10%), 60 Hz, 170 VA(max.)

Operating temperature

2 °C – 35 °C at max. 80% relative
humidity

Storage temperature

5 °C – 45 °C at max. 80% relative
humidity

Water connection input
1 × ⅜” PLC quick-connect coupling

Water connection output
Bagtank 20

1 × ⅜” PLC quick-connect coupling

Bagtank 50, Bagtank 100

2 × ⅜” PLC quick-connect coupling1 Bagtank 20 is supplied without a pump as standard, pump optionally available
2 Value only applies to Bagtank 20, dispensing site at the same height or lower than the tank outlet
3 Note: The Arium® Bag is not included in the scope of delivery of the Arium® Bagtank

6Arium® Smart Station---Remote dispensing at high flexibility
Compact: Save space integrating in your lab
Intuitive: Touch-activated color display with direct accessto all important functionsFlexible: Stepless height adjustment to fill different size
containers
Accurate: Precise volume dispense for reliable bufferand sample preparation

Description
The Arium® Smart Station is designed for flexible remote
dispensing of pure water directly at the point of use. Whiledispensing water into a broad range of different sized
containers, the Smart Station offers constant control ofevery important quality parameters, at all times. The
ergonomic design supports left-and right-hand operationand can be easily adjusted to your need.
Using the extended connection set, the distance between
the Smart Station and the Arium® Bagtank can be set up to4 meters. Based on your needs different point of use filters
can be added.

Arium® Smart Station Pure:
Supply pure water from Arium® Bagtank

7Technical Specifications |Ordering Information
Dimensions Smart Station Bench-Top

Order number

Description

Control box with stand(w × d × h)

213 × 213 × 598 mm(8.4 × 8.4 × 8.2”)

H2O-ARST-P-T

Arium® Smart Station Purefor benchtop installation

Operating range fixed
dispense arm (d × w × h)

428 × 476 × 835 mm(16.9 × 18.7 × 32.9”)

H2O-ARST-P-B

Arium® Smart Station Pure forwall-mounted installation

Tubing Length:
Distance to water system |bagtank

2 Meter

Operating range flexible hand
held0.7 Meter

WeightApprox. 4.9 kg (10.8 Ibs)

Dimensions Smart Station Wall-Mounted
Control box (w × d × h)

172 × 157 × 343 mm(6.8 × 6.2 × 13.5”)

Operating range
dispense arm (d × w × h)

242 × 90 × 300 mm(9.5 × 3.5 × 11.8”)

Tubing Length:
Distance to water system |bagtank

2 Meter

Operating range flexible hand
held0.7 Meter

WeightApprox. 2.4 kg (5.3 Ibs)

General Specifications
Volume-controlled dispensing

0.05 – 50 L in 50 mL steps

Volume accuracy±5% between

Power supply

100 – 240 VAC; 50 and 60 Hz,
2.5 A (max.) 2 °C – 40 °C

Power cord (IEC 60320-1 /C14)

Country specific

8Benchtop and wall-mounted edition can be assembeled forleft or right hand side, without additional equipment required.
Accessories and final filters for the Arium® Smart Station canbe found in the Arium® Smart Station datasheet.

-Intended Use
Device type:Arium® Bagtank

Arium® Water Guard-----

Early detection of leakages protects the laboratory
Highly sensitive optical sensor
Audiovisual alarm signals
Automatic water stop in the case of leakage
High-quality material, no corrosion
Easy to install
Integrated wall mounting bracket for solenoid valve

Description
Only the early detection of water leakages provides optimal
protection against water damage in the laboratory.
Leak­ages are registered by the highly sensitive opticalsensor.
In contrast to conventional sensors, this sensor functions
independently of conductivity measurement values asthese are so low in the pure water that the activation of theguard would not be guaranteed. Once a leak is detected,
the water guard automatically locks the feed water inlet
line. An acoustic warning is triggered immediately and thesystem status can be constantly controlled using theintegrated LED display. With its sensitive optical sensors andhigh-quality materials, the Arium® Water Guard is perfectfor all pure water systems.

Technical Specifications |Ordering Information
Sensor dimensions
Diameter

5 cm

Height2.5 cmCable length2m

Tubing connections
Input

⅜” Plug-in connector

Output⅜” Plug-in connector

Power supply

100 – 240 VAC50 – 60 Hz

Order number

Description

610AWG1Arium® Water Guard, 1 pc

-Intended Use
Device type:Arium® Advance EDI

9Arium® RO Pretreatment Cartridge---Safe protection of the RO module
Fast and effective adsorption of impurities throughhigh-grade activated carbon
5 µm depth filter for the retention of particles
Highly efficient catalyst for removing free chlorine
Patented cartridge design for easy installation

Description
An efficient protection for a downstream reverse osmosis(RO) membrane is the combination of spherical, catalytic
effective activated carbon, a catalyst and a depth filter.
It reliably removes oxidation agents, such as free chlorine,
heavy-metal ions and particulate contaminants from thefeed water of the system.
A special catalyst is an integral part of pre-treatment. It isparticularly efficient at removing free chlorine and at a lower
temperature and/or higher pH value compared to activatedcarbon alone.
The patented cartridge design ensures minimal time
expenditure with ultra-easy installation and exchange.

Technical Specifications |Ordering Information
Materials
HousingHigh-quality polypropylene

Cleaning media

Spherical catalytic effective activatedcarbon plus polypropylene filter cartridge with nom. 5 µm separation rate

Dimensions [W × H × D]

18 × 26 × 11 cm

Operating weight

3.5 kgFeed water requirements

See “Technical Specifications” page 2

10Order number

Description

H2O-CPFAD-1

Arium® RO Pretreatment Cartridge,
1 pc-Intended Use
Device type:
Arium® Advance EDI

Arium® RO Modules----

Reverse osmosis modules with low-energy membranes
Highly efficient reverse osmosis membranes, optimized
water consumption
Low-energy membranes for ecological and economical
operation
Backflush with product water increases the service life
Easy replacement
Constant flow
Consistently high water quality

Description
The Arium® RO modules consist of two independent
membranes whose design guarantees easy installation
and reliable operation. Each of the two modules contains
a low-energy reverse osmosis membrane in a
polypro­pylene housing.
The housing has connections for feed water, permeate(product water) and concentrate (discarded water). The
RO modules enable an ideal water yield, thereby
optimizing the water consumption. At the same time, up
to 98% of the salts are typically retained. Thanks to thebackflush with permeate, particles and salts are removed
from the surface of the membrane.

This results in a longer service life and lower system
maintenance costs. In addition, this backflush function on
restarting the system after a standstill allows for theimmediate dispensing of high quality water.

Technical Specifications |Ordering Information
Materials
RO membranes

Low-energy membrane madeof polyamide

Housing

Polypropylene

Dimensions for each module
Height30.8 cm

Diameter

7.8 cmWeight0.468 kg

Product Water Quality

See “Technical Specifications” page 2

Order numberDescription

H2O-CRO-H- 1Arium® RO Module, 1 pc

H2O-CRO-H- 2Arium® RO Module, 2 pcs-Intended Use
Device type:
Arium® Advance EDI11Arium® Softener Cartridge---For maximum service life of the EDI module
Consistently high water quality
Long service life
Effective CaCO3 elimination

Description
It is sensible to soften the feed water to improve protectionof the EDI module.
The cartridge reliably removes traces of alkaline earth ionsfrom the water, thereby guaranteeing consistently highwater quality and a long service life of the EDI module.

Technical Specifications |Ordering Information
MaterialsHousing

High-purity polypropylene

Filling material

Ultrapure ion exchange resin

Order number

Description

H2O-CSO-1

Arium® Softener Cartridge, 1 pc

-Intended Use
Device type:Arium® Advance EDI

12Arium® RO Cleaning Set----

Maximum service life of the RO module
Effective removal of scaling and metal deposits
Elimination of organic compounds
Dispersion of colloids
Stable pH values
Gentle on materials

Description
Two-stage cleaning kit for removing scaling and organic
impurities.
The alkaline substance contains non-foaming surfactants
that dissolve organic compounds, disperse colloids andcan be quickly removed again from the membrane surface.
Cleaning efficiency depends on the pH value that is
steadily maintained by buffer substances through a large
temperature range.
The acidic cleaning agent to remove scaling contains
chelate and reducing agents in order to dissolve metallicdeposits. The ideal pH value also remains consistently
low over a wide range during cleaning in this case thanks
to the buffers.

Technical Specifications |Ordering Information
Ingredients

Order numberDescription
Arium® RO Cleaning Set, 1 pc

Alkaline cleaner

HEDTA, ethanolamine, triethanolamine

H2O-CCS

Acidic cleaner

HEDTA, phosphoric acid, citric acid

Intended Use
Device type:
Arium® Advance EDI-13

Sartorius ServiceWe Ensure the Quality of Your Results

Our Services at a Glance:

At Sartorius, quality products go hand in hand withprofessional service. With our wide service offering, we will
help guarantee the reliable and optimal operation of yourArium® systems. Just ask us and we will even cover theentire life cycle of your laboratory water system – fromcommissioning to qualification to regular maintenance.
Together with you, we will ensure the consistently highquality of your laboratory water purification.

Installation and Commissioning
Your advantage: Your system will operate reliably at peak
performance from day one
Equipment Qualification (IQ | OQ)Your advantage: You will meet all regulatory requirements(GMP | GLP)
Regular Preventative Maintenance, Including Calibration,
inspection and testing of your system and exchange ofconsumables
Your advantages: Optimal operation of your system; reliableresults; prevention of downtime or even equipment failure
Get more information now at:www.sartorius.com/en/services

GermanySartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Straße 2037079 Göttingen
Phone +49 551 308 0F or further information, visit
www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 04 | 2023

USASartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906