Product Datasheet

Arium®
Advance RO
Reverse Osmosis System
With iJust--Advantages
Time-saving – Use of innovative bag technology,
eliminates costly tank cleaning
Optimized water consumption – Automatic with iJust
Easy to use – Display with touch function
and intuitive menu

Product Description
The Arium® Advance RO provides high-quality Type 3 reverse
osmosis water. By using the integrated, innovative iJust
function, it does not only optimizes water consumption but
also ensures that natural resources are used conscientiously.
In contrast to conventional water purification systems, theunique touch display with intuitive menu navigation makes it
extremely easy to use.
With a flow rate of up to 8, 16 or 24 L/h, the automatic RO
membrane backflush and a constant flow rate, the Arium®
Advance RO is the optimal choice for general laboratory
applications.

Innovative Bag Technology

iJust

The pure water is stored in the enclosed Arium® Bagtanksystem. This guarantees optimal storage of the pure waterand protects against secondary contamination. Timeconsuming tank cleaning intervals are eliminated thanks to
the interchangeable bag.

iJust stands for innovative technology that optimizes waterproduction. Amongst other things, the intelligent Arium®
software controls a valve on the concentrate outlet inaccordance with the data that have been entered for
CaCO₃ and CO₂. iJust therefore optimizes the productwater quality and water consumption.

Display With Touch Function
Simply navigate intuitively in the easy-to-use and clear
menu by lightly touching the display – even with gloves.

---High product water quality at all times
Optimized, economical water consumption
Guarantees a longer life of the downstream ultrapure
water systems

Technical SpecificationsDimensions [W × H × D]

35.0 × 50.1 × 45.1 cm

Empty weightapprox. 15 kg

Operating weightapprox. 22 kg

Power supply

100 – 240 VAC (±10%); 50 – -60 Hz, 130 VA (max.)

Operating temperature

2 °C – 35 °C at max. 80% relative humidity

Storage temperature

5 °C – 45 °C at max. 80% relative humidity

Feed Water Quality
Exclusively potable tap water pursuant to the drinking water standards of the USA, the European Union, or Japan.
Input pressure10.5 – 6.9 bar, recommended > 2 bar

Temperature2 – 30 °CSpecific conductivity< 1,500 µS/cm compensated to 25 °C

TOC< 2,000 ppb

Max. total hardness (max. CaCO3)

360 ppm

Free chlorine< 4 ppm

Iron (total Fe content)< 0.1 ppm

Fouling Index (SDI)<5Turbidity< 1 NTU

pH value4 – 10

1 Dynamic pressure | flow pressure 100 L/h2

RO module #1

Conductivity measurement(product water | permeate) (RO)Conductivitymeasurement(feedwater)

RO module #2
Inlet(feed water)

RO pretreatment
cartridge

Pump

Outlet(waste water)

Arium®
Bagtank

Flow chart Arium® Advance RO (H2O-RO-2-T)

Water Applications
Advance ROWater Quality
Type 3 Reverse Osmosis Water

Lab Water System by Daily Water Consumption
Type 3 reverse osmosis water approx. 140 Liter/day (8 L/h)
Type 3 reverse osmosis water approx. 200 Liter/day (16 L/h)
Type 3 reverse osmosis water approx. 270 Liter/day (24 L/h)Feed Application
Feed ultrapure water systems
Water for Laboratory devices (Autoclaves | Washing Machine etc.)

General Laboratory Application
Buffer, media and pH solutions---3

Product Water Quality
Water purification method

Particle filtration, adsorption by means of spherical activated carbon, catalyst and reverse osmosis

Water typeType 3, reverse osmosis water

Production output²

8, 16 or 24 L/hWater dispensing flow rate³

Up to 3 L/min

Typical conductivity1< 20 µS/cm

Typical resistivity1> 0.05 MΩ × cm

Typical ion retention

up to 98%

Retention of dissolved organic
substances (MW > 300 Dalton)> 99%

Particle and microorganism retention > 99%¹ Depending on the feed water quality and temperature²D epending on the feed water pressure, temperature, and condition of the RO modules³ When using an Arium® Bagtank with pump, depending on hydrostatic pressure, connected accessories or end filter

Additional product water specifications when connected to an Arium® Smart Station¹ with final filter
Particle content²

No particles > 0.22 µm

Bacteria²< 0.001 CFU/mL

Endotoxins³< 0.001 EU/mL

RNase concentration³< 1 pg/mL

DNase concentration³< 5 pg/mL

Water dispensing flow rate⁴

Up to 2 L/min

Volume-controlled dispensing

0.05 – 50 L in 50 mL steps

1 Connected to an Arium® Bagtank² When using an Arium® Sterile Plus (Sartopore® 2 150 final filter)³ When using an Arium® Cell Plus final filter
⁴D epending on the connected Arium® Bagtank, hydrostatic pressure, connected accessories and end filter
Accessories and final filters for the Arium® Smart Station can be found in the Arium® Smart Station data sheet.

Ordering Information
Arium® Advance RO reverse osmosis systems for the production of Type 3 pure water
Scope of supply: 1 Arium® Advance RO, RO-(reverse osmosis) module(s), connection kit and water guard
Order number

Description

H2O-RO-1-T

Arium® Advance RO benchtop device, flow capacity 8 L/h

H2O-RO-1-B

Arium® Advance RO wall-mounted device, flow capacity 8 L/h

H2O-RO-2-T

Arium® Advance RO benchtop device, flow capacity 16 L/h

H2O-RO-2-B

Arium® Advance RO wall-mounted device, flow capacity 16 L/h

H2O-RO-3-T

Arium® Advance RO benchtop device, flow capacity 24 L/h

H2O-RO-3-B

Arium® Advance RO wall-mounted device, flow capacity 24 L/h

4Accessories
Arium® Bagtank---

Innovative tank system
Integrated ventilation filter with non-return valve
provides reliable protection against CO2 pollution
High flexibility through the 4 rollers available as an option
Easy and fast exchange of the Arium® Bags
No chemical cleaning required

Description
The pure water is stored in the innovative enclosed Arium®
Bagtank system. This system protects the prepared purewater against secondary contamination. The Sartorius
Bagtank system enables consistent water quality over a
prolonged period, thereby ensuring permanent,reproducible results. Unlike conventional water reservoirs,
the Arium® Bag eliminates time consuming tank cleaningintervals thanks to the interchangeable technology.
Arium® Bagtanks are housings which are equipped withArium® Bags. The Arium® Bagtanks are available in 20 L, 50 L,
and 100 L volumes. Their design is adaptable and saves
space in any laboratory environment, and the optionalrollers make this an extremely flexible system.
Integrated distributor pumps are a standard componentof the 50 L and 100 L Bagtanks. A distributor pump is alsoavailable as an option for the 20 L Bagtank. In addition, awall holder for the space-saving and user-friendly
installation of this tank is also available.

Water dispensing flow rate
With pump¹

up to 3.0 L/min

With pump, remote dispenser andSterile Plus filter¹

up to 2.0 L/min

Without pump²

up to 1.5 L/min

Pump pressure

3 bar

1 Bagtank 20 is supplied without a pump as standard, pump optionally available
2 Value only applies to Bagtank 20, dispensing site at the same height or lower than the tank outlet-

Intended Use
Device type:
Arium® Advance RO5

Technical Specifications |Ordering Information
Order number

Description

H2O-AOV-20³

Arium® Bagtank 20 L, without pump,
1 pc

H2O-AOV-50³

Arium® Bagtank 50 L, with pump
240 VAC, 50 Hz, 1 pc

H2O-AOV-50-US³

Arium® Bagtank 50 L, with pump
115 VAC, 60 Hz, 1 pc

H2O-AOV-50-W³Arium® Bagtank 50 L, without pump,
1 pc

H2O-AOV-100³Arium® Bagtank 100 L, with pump
240 VAC, 50 Hz, 1 pc

H2O-AOV-100-US3Arium® Bagtank 100 L, with pump
115 VAC, 60 Hz, 1 pc

H2O-AOV-100-W3Arium® Bagtank 100 L, without pump,
1 pc

Empty weight without Arium® Bag |Operating weight with filled Arium® Bag

H2O-ADP-20

Pump Arium® Bagtank 20 L, 240 VAC,50 Hz, 1 pc

Bagtank 20

19 kg | 40 kg

H2O-ADP-20-US

Bagtank 5033 kg | 84 kg

Pump Arium® Bagtank 20 L, 115 VAC,60 Hz, 1 pc

Bagtank 100

47 kg | 148 kg

H2O-ATR

Rollers for Arium® Bagtank 50 & Bagtank 100, including fastening material,
4 pcs

H2O-CBS-20

Arium® 20 L Bag for Arium® 20 L
Bagtank, 2 pcs

H2O-CBS-50

Arium® 50 L Bag for Arium® 50 L and100 L Bagtank, 2 pcs

H2O-ATB

Wall mount for Arium® Bagtank 20,
1 pc

Materials
Bagtank

Stainless steel | plastic

Bag

S71 filmTubing

PE | silicone

Dimensions, excluding rollers and wall bracket [H × W × D]
Bagtank 20

80.8 × 16.6 × 43.7 cm

Bagtank 50

85.2 × 25.4 × 58.7 cm

Bagtank 100

85.2 × 51.4 × 58.7 cm

Bag 20 L

86.5 × 43.0 cm

Bag 50 L

90.0 × 58.1 cm

Number of Bags per tank
Bagtank 20

1 × 20 L

Bagtank 50

1 × 50 L

Bagtank 100

2 × 50 L

Power supply1

240 VAC (±10%), 50 Hz, 120 VA (max.)

Power supply
US versions1

115 VAC (±10%), 60 Hz, 170 VA(max.)

Operating temperature

2 °C – 35 °C at max. 80% relative
humidity

Storage temperature

5 °C – 4 °C at max. 80% relative
humidity

Water connection input
1 × ⅜” PLC quick-connect coupling

Water connection output
Bagtank 20

1 × ⅜” PLC quick-connect coupling

Bagtank 50, Bagtank 100

2 × ⅜” PLC quick-connect coupling1 Bagtank 20 is supplied without a pump as standard, pump optionally available
2 Value only applies to Bagtank 20, dispensing site at the same height or lower than the tank outlet
3 Note: The Arium® Bag is not included in the scope of delivery of the Arium® Bagtank

6Arium® Smart Station---Remote dispensing at high flexibility
Compact: Save space integrating in your lab
Intuitive: Touch-activated color display with direct accessto all important functions
Flexible: Stepless height adjustment to fill different size
containers
Accurate: Precise volume dispense for reliable bufferand sample preparation

Description
The Arium® Smart Station is designed for flexible remote
dispensing of pure water directly at the point of use. Whiledispensing water into a broad range of different sized
containers, the Smart Station offers constant control ofevery important quality parameters, at all times. The
ergonomic design supports left-and right-hand operationand can be easily adjusted to your need.
Using the extended connection set, the distance between
the Smart Station and the Arium® Bagtank can be set up to4 meters. Based on your needs different point of use filters
can be added.

Arium® Smart Station Pure:
Supply pure water from Arium® Bagtank

7

Technical Specifications |Ordering Information
Dimensions Smart Station Bench-Top
Control box with stand(w × d × h)

213 × 213 × 598 mm(8.4 × 8.4 × 8.2”)

Operating range fixeddispense arm (d × w × h)

428 × 476 × 835 mm(16.9 × 18.7 × 32.9”)

Tubing Length:
Distance to water system |bagtank

2 Meter

Operating range flexible hand
held0.7 Meter

WeightApprox. 4.9 kg (10.8 Ibs)

Dimensions Smart Station Wall-Mounted
Control box (w × d × h)

172 × 157 × 343 mm(6.8 × 6.2 × 13.5”)

Operating range
dispense arm (d × w × h)

242 × 90 × 300 mm(9.5 × 3.5 × 11.8”)

Tubing Length:
Distance to water system |bagtank

2 Meter

Operating range flexible hand
held0.7 Meter

WeightApprox. 2.4 kg (5.3 Ibs)

General Specifications
Volume-controlled dispensing

0.05 – 50 L in 50 mL steps

Volume accuracy±5% between

Power supply

100 – 240 VAC; 50 and 60 Hz,
2.5 A (max.) 2 °C – 40 °C

8Power cord (IEC 60320-1 /C14)

Country specificOrder number

Description

H2O-ARST-P-T

Arium® Smart Station Purefor benchtop installation

H2O-ARST-P-B

Arium® Smart Station Pure forwall-mounted installation

Benchtop and wall-mounted edition can be assembeled forleft or right hand side, without additional equipment required.
Accessories and final filters for the Arium® Smart Station canbe found in the Arium® Smart Station data sheet.

-Intended Use
Device type:
Arium® Bagtank

Arium® Water Guard-----

Early detection of leakages protects the laboratory
Highly sensitive optical sensor
Audiovisual alarm signals
Automatic water stop in the case of leakage
High-quality material, no corrosion
Easy to install
Integrated wall mounting bracket for solenoid valve

Description
Only the early detection of water leakages provides opti­mal
protection against water damage in the laboratory.
Leak­ages are registered by the highly sensitive opticalsensor.
In contrast to conventional sensors, this sensor functions
independently of conductivity measurement values, asthese are so low in the pure water range that the activationof the guard would not be guaranteed. Once a leak isdetected, the water guard automatically locks the feedwater inlet line. An acoustic warning is triggered immedi­
ately and the system status can be constantly controlledusing the integrated LED display. With its sensitive opticalsensors and high-quality materials, the Arium® Water Guardis perfect for all pure water systems.

Technical Specifications |Ordering Information
Sensor dimensionsDiameter

5 cm

Height2.5 cmCable length2mTubing connections
Input

⅜” Plug-in connector

Output⅜” Plug-in connector

Power supply

100 – 240 VAC 50 – 60 Hz

Order numberDescription

610AWG1Arium® Water Guard, 1 pc-

Intended Use
Device type:
Arium® Advance RO

9Arium® RO Pretreatment Cartridge---

Safe protection of the RO module
Fast and effective adsorption of impurities throughhigh-grade activated carbon
5 µm depth filter for the retention of particles
Highly efficient catalyst for removing free chlorine
Patented cartridge design for easy installation

Description
An efficient protection for a downstream reverse osmosis(RO) membrane is the combination of spherical, catalytic
effective activated carbon, a catalyst and a depth filter.
It reliably removes oxidation agents, such as free chlorine,
heavy-metal ions and particulate contaminants from thefeed water of the system.
A special catalyst is an integral part of pre-treatment. It isparticularly efficient at removing free chlorine and at a lower
temperature and/or higher pH value compared to activatedcarbon alone.
The patented cartridge design ensures minimal time
expenditure with ultra-easy installation and exchange.

Technical Specifications |Ordering Information
Materials
HousingHigh-quality polypropylene

Cleaning media

Spherical catalytic effective activatedcarbon plus polypropylene filter cartridge with nom. 5 µm separation rate

Dimensions [W × H × D]

18 × 26 × 11 cm

Operating weight

3.5 kgFeed water requirements

See “Technical Specifications” page 2

10Order number

Description

H2O-CPFAD-1

Arium® RO Pretreatment Cartridge,
1 pc-Intended Use
Device type:Arium® Advance RO

Arium® RO Modules----Reverse osmosis modules with low-energy membranes
Highly efficient reverse osmosis membranes, optimized
water consumption
Low-energy membranes for ecological and economical
operation
Backflush with product water increases the service life
Easy replacement
Constant flow
Consistently high water quality

Description
The Arium® RO modules consist of two independent
membranes whose design guarantees easy installation
and reliable operation. Each of the two modules contains
a low-energy reverse osmosis membrane in a polypro­
pylene housing.
The housing has connections for feed water, permeate(product water) and concentrate (discarded water). The
RO modules enable an ideal water yield, thereby
optimizing the water consumption. At the same time, up
to 98% of the salts are typically retained. Thanks to thebackflush with permeate, particles and salts are removed
from the surface of the membrane.

This results in a longer service life and lower system
maintenance costs. In addition, this backflush function on
restarting the system after a standstill allows for theimmediate dispensing of high quality water.

Technical Specifications |Ordering Information
Materials
RO membranes

Low-energy membrane madeof polyamide

Housing

PolypropyleneOrder number

Description

613CPM4Arium® RO Module, 1 pc

613CPM4--------V

Arium® RO Module, 2 pcs-Intended Use
Device type:
Arium® Advance RO

Dimensions for each module
Height30.8 cm

Diameter

7.8 cmWeight0.468 kg

Product Water Quality

See “Technical Specifications” page 211

Arium® RO Cleaning Set----

Maximum service life of the RO module
Effective removal of scaling and metal deposits
Elimination of organic compounds
Dispersion of colloids
Stable pH values
Gentle on materials

Description
Two-stage cleaning kit for removing scaling and organic
impurities.
The alkaline substance contains non-foaming surfactants
that dissolve organic compounds, disperse colloids andcan be quickly removed again from the membrane surface.
Cleaning efficiency depends on the pH value that is
steadily maintained by buffer substances through a large
temperature range.
The acidic cleaning agent to remove scaling contains
chelate and reducing agents in order to dissolve metallic
deposits. The ideal pH value also remains consistently
low over a wide range during cleaning in this case thanks
to the buffers.

Technical Specifications |Ordering InformationIngredients
Alkaline cleaner

HEDTA, ethanolamine, triethanolamine

Acidic cleaner

HEDTA, phosphoric acid, citric acid12

Order number

Description

H2O-CCSArium® RO cleaning Set, 1 pc-

Intended Use
Device type:Arium® Advance ROSartorius Service
We Ensure the Quality of Your Results

Our Services at a Glance:

At Sartorius, quality products go hand in hand withprofessional service. With our wide service offering, we will
help guarantee the reliable and optimal operation of yourArium® systems. Just ask us and we will even cover theentire life cycle of your laboratory water system – fromcommissioning to qualification to regular maintenance.
Together with you, we will ensure the consistently highquality of your laboratory water purification.

Installation and Commissioning
Your advantage: Your system will operate reliably at peak
performance from day one
Equipment Qualification (IQ | OQ)Your advantage: You will meet all regulatory requirements(GMP | GLP)
Regular Preventative Maintenance, Including Calibration,
inspection and testing of your system and exchange ofconsumables
Your advantages: Optimal operation of your system; reliableresults; prevention of downtime or even equipment failure
Get more information now at:www.sartorius.com/en/services

13Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0F or further information, visit
www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 04 | 2023

USASartorius Corporation565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906