series

100120S/D1 and 120U/D1 cased pumps
fitted with 400D1 pumphead

8BUTPO.BSMPX1VNQT(SPVQFEATURES AND BENEFITSt'MPXSBUFTUPNMNJOBUCBS QTJ
t$PNQBDUGPPUQSJOUSFRVJSJOHNJOJNBMCFODIUPQTQBDFt4VQFSJPSTQFFEDPOUSPMVQUP  TFUTQFFEBDDVSBDZt%.JTBUXPDIBOOFMQVNQIFBEBOE%.BUISFFDIBOOFMQVNQIFBECPUIGPSEPVCMFTFHNFOUNBOJGPMEUVCJOHt*1JOHSFTTQSPUFDUJPOGPSXJQFEPXOFOWJSPONFOUTt.BJOUFOBODFGSFFCSVTIMFTT%$NPUPSTECHNICAL DATA'VODUJPOBMJUZ

120S120U.BOVBMDPOUSPM
t

1 to 200rpm (200:1)t

0.1 to 200rpm (2000:1)
Numerical speed displayttReversible directiontt

Max (prime)ttAuto starttt

3FNPUFDPOUSPMt

Run/stop, direction change, auto/man toggle, alarm"OBMPHVFDPOUSPM
0.01 to 200rpm (20,000:1)

t

Analogue speed input; 4-20mA, 0-10VtTacho speed output; 0-5Vt

4FDVSJUZt

Keypad lock to protect set-uptPERFORMANCE
5VCFCPSFBOEþPXSBUFT4%BOE6% NMNJO
Speed rpm

NN

NN

NN

NN

NN

NN

120S/D11 - 200rpm

0.01 – 2.2

0.03 – 5.8

0.11 – 23

0.24 – 49

0.41 – 81

0.59 – 120

120U/D1

0.1 - 200rpm

0.001 – 2.2

0.003 – 5.8

0.011 – 23

0.024 – 49

0.041 – 81

0.059 - 120

DIMENSIONSTECHNICAL SPECIFICATION

MATERIALS OF CONSTRUCTION

4 6$PNQPOFOU.BUFSJBM

Ingress rating

IP31 / NEMA 2Keypad

Polyester

Enclosure

Wipedown

Drive casework

ABS plastic

Suitability

Dry light-industrial environments

Drive shaft

Stainless SteelHumidity

Non-condensing 80% up to 31C, 88F, decreasing linearly
to 50% at 40C, 104F

Pumphead track

Aluminium (anodised)

Rotor

Aluminium (anodised)

Temperature

5C to 40C (40F to 104F)

Rollers

AcetalPump weight

1.65kg (3lbs 9oz)

Spindles guide pins

Stainless steel

Control ratio

120S: 200:1, 120U: Manual 2000:1, Auto 20,000:1

Torque rating

0.4Nm continuous duty

Noise

<60dB(A) at 1m

StandardCE, cETLus

Power supply

24VDC from mains transformer supplied(mains supply range 90~264VAC 47~63Hz)

ORDER INFORMATION

1VNQBOEQVNQIFBEQSPEVDUDPEFT%FTDSJQUJPO

1BSUDPEF%FTDSJQUJPO

1BSUDPEF

120S/D1 manual speed pump

010.6131.010

400D1 pumphead

043.001D.D1C

120U/D1 manual and auto pump

010.6141.010

5VCJOHNNXBMMUIJDLOFTTGPS%
NN

NN

NN

NN

NN

#JPQSFOF

933.0005.016

933.0008.016

933.0016.016

933.0024.016

933.0032.016

1VNQTJM 4JMJDPOF

913.A005.016

913.A008.016

913.A016.016

913.A024.016

913.A032.016

902.0024.016

902.0032.016

Marprene

902.0005.016

902.0008.016

45"163&1'-/FPQSFOF

920.0008.016

902.0016.016

961.0032.016

966.0016.016

966.0032.016

920.0016.016

920.0032.016

902.0040.016※１．内径4.0mmのチューブは廃番になりました

All flow rates shown were obtained pumping water at 20C (68F) with zero suction and delivery heads. Watson-Marlow,Bioprene and Marprene are trademarks of Watson-Marlow Limited. Disclaimer: The information contained in thisdocument is believed to be correct but Watson-Marlow Limited accepts no liability for any errors it contains, andreserves the right to alter specifications without notice. LoadSure is a trademark of Watson-Marlow Limited.
® STA-PURE PFL and ® STA-PURE PCS are registered trademarks of W.L Gore & Associates Inc.
Please state the product code when ordering pumps and tubing.

wmpg.co.uk
info@wmpg.co.uk+44 (0) 1326 370370

wd-120sdud-gb-02 © Copyright 2013 Watson-Marlow Pumps Group

961.0016.016

45"163&1$4

NN ※１