Professional Blood Pressure
Monitors UM-Series

UM-102The UM-102A is a Mercury free sphygmomanometerfrom A&D. It is a reliable and accurate manual
sphygmomanometer, with an ergonomic, modern andlightweight design. Designed for professional use, it
allows clinicians to record accurate blood pressure
measurements using the "Gold Standard" Auscultatory
measurement technique whilst listening for Korotkoff
sounds with a stethoscope.
Auscultatory blood pressure measurement
Digital readout for pulse rate
Cuff holder and grip for carrying
Durable and chemical resistant body
Full ABS housing light weight body
Professional cuff and 5 cuff sizes are available for use
Adjustable display column for easier analysis of result
Easy battery replacement
Anti-glare display minimizes reflected lightA-type Traditional Style

Easy display adjustment

B-type New Dual Use Design

Flat base for stacking
multiple units

A-typeEasy battery replacement

In combination
with optional mobile standB-typeStand-alone use

Specifications
UM-102

Measurement
methodAuscultatory measurement with stethoscope

Measurement
range

Pressure: 0 mmHg - 299 mmHg
Pulse: 40 bpm - 180 bpm

Measurement
accuracy

Pressure: ±3 mmHg, Pulse: ±5 %Power supply

2 x 1.5 V Batteries (LR6 or AA)

Battery life

Approx. 2,000 measurements

Memory storage

N/A

Operatingconditions+10 °C to 40 °C / 15 %RH to 85 %RH /800 hPa~1060 hPa

Transport /Storage conditions-20 °C to +60 °C / 10 %RH to 95 %RH

Dimensions

Approx. 98<W> x 67<H> x 324<D> mm(A-type)
Approx. 98<W> x 326<H> x 202<D> mm(B-type)

WeightApprox. 520 g (Unit without Battery)(A-type)
Approx. 540 g (Unit without Battery)(B-type)

OptionREF:

Option Name

CUF-KW-LL

LL Cuff (41 cm - 50 cm)

CUF-KW-LA

Large Adult Cuff (31 cm - 45 cm)

CUF-KW-A

Adult Cuff (22 cm - 32 cm)

CUF-KW-SA

Small Adult Cuff (16 cm - 24 cm)

CUF-KW-SS

SS Cuff (12 cm - 17 cm)

UM-102-10

Rubber Bulb Unit

UM-102-11

Connector and Plugs forCuff Replacement

UM-ST001

Mobile Stand (for B-type only)

External Dimensions(mm/inches)
B-type with optional stand910 ~ 1260 / 35.8 ~ 50.0

Model Name

Standard accessories

Adultcuff

Rubber BulbUnit

2 x AABatteries

400 / 15.7

Mercury-Free

UM-211

Professional Blood Pressure MonitorsThe UM-211 automated office blood pressure monitor
provides your office with a consistent, highly accurate
and reliable blood pressure measurement tool. Itsfeatures and durability are designed to meet the needsof in-clinic or office settings.
Dual measurement mode (Auscultatory and Oscillometric)
Cuff holder and grip for carrying
Durable and chemical resistant body
Professional cuff and 5 cuff sizes are available for use
Large LCD display with backlight

Auscultatory mode

Rechargeable battery (300 measurements at full charged)
Adjustable inflation pressure (AUTO/220/250/280)

UM-Series’ auscultatory mode is designed for daily
and frequent use. Selectable deflation speed (2.5mmHg / sec. or 5.0 mmHg / sec.) and quick start for
auscultatory mode are features `Only from A&D'.

Irregular heartbeat (IHB) indicator
Room temperature display

Quick Start

Just pressing START + MODE button.

Selectable Deflation Speed
Stand-by display(Room temperature)

Measurement display(with backlight)

2.5 mmHg / Sec. or
5.0 mmHg / Sec.

Easy to carry

Specifications
UM-211

OptionMeasurement
methodOscillometric measurement andAuscultatory measurement withstethoscope

REF:

Option Name

UM-CUF-KS-LL

LL Cuff (41 cm - 50 cm)

Measurement
range

UM-CUF-KS-LA

Large Adult Cuff (31 cm - 45 cm)

Pressure:0 mmHg - 299 mmHg
Pulse:40-200 beats / minute

Measurement
accuracy

UM-CUF-KS-AU

Adult Cuff (22 cm - 32 cm)

UM-CUF-KS-SA

Small Adult Cuff (16 cm - 24 cm)

Pressure:±3 mmHg, Pulse:±5 %UM-CUF-KS-SS

SS Cuff (12 cm - 17 cm)

UM-211-20

Battery Pack

Power supply

AC 100 V - 240 V (Rechargeable Battery)TB-268AC Adapter

Battery life

Approx. 300 measurements

KO1886/7/8/9

AC Cable (A/C/BF/O Plug)

UM-ST002

Mobile Stand for UM-211

Memory storage

Last 99 measurements

Operatingconditions+5 ˚C to 40 ˚C /10 %RH to 85 %RH /800 hPa to 1060 hPa

Standard accessories

Transport /Storage conditions-20 ˚C to +60 ˚C / 10 %RH to 95 %RH

Adultcuff

BatteryPack

Dimensions

Approx. 120<W> x 200<H> x 140<D> mm

WeightApprox. 550 g (Unit without Bat tery)
AC Adapter

AC CableExternal dimensions(mm/inches)with optional stand UM-ST002

750 ~ 1100 / 29.5 ~ 43.3

Model Name

Option cuff (5 sizes)

400 / 15.7

A&D Company Limited.(JAPAN)

A&D AUSTRALASIA PTY LTD(Australia)

A&D ENGINEERING, INC.(USA)A&D RUS CO., LTD.(Russia)

A&D Instruments Canada Inc.(CANADA)

A&D Technology Trading(Shanghai)Co., Ltd(China)

A&D INSTRUMENTS LIMITED(United Kingdom)

A&D INSTRUMENTS INDIA PRIVATE LIMITED(India)

URL:aandd.jp

URL:andonline.com

URL:autocontrol.com

URL:andprecision.com

URL:andaustralasia.com.au
URL:and-rus.ru

URL:aanddtech.cn
URL:aanddindia.in*UM102+UM211-ADCC-01-AD01-245001GP