SAZ-2410-19-S1

Rev. 1.5

Description:
Model SAZ-2410-19-S1 is a U-band standard gainhorn antenna that operates from 40 GHz to 60 GHz.
The antenna offers 24 dBi nominal gain, a typical half
power beamwidth of 9.7 degrees on the E-plane and11.0 degrees on the H-plane at the center frequency,respectively. The antenna supports linear polarized
waveforms. The input of this antenna is a WR-19
waveguide with UG-383/U-M anti-cocking flange. Thestandard gain horn is offered for antenna range
calibration purpose mainly, but it can be also used for general purpose system set ups.

Features:••••Applications:••••

Rectangular Waveguide Interface
Precisely Machined and Gold Plated
Linear Polarization
High Return Loss

Antenna Range Standards
Antenna Gain Measurements
General System Setups
Radar/Communication Systems

Electrical Specifications:Parameter
Frequency
Gain
Polarization
3 dB Beamwidth, E-Plane @50GHz
3 dB Beamwidth, H-Plane @50GHz
Sidelobes, E-Plane
Sidelobes, H-Plane
Return Loss
Specification Temperature
Operation Temperature

Minimum
40 GHzTypicalSAGE Millimeter, Inc.

WR-19 Standard Gain Horn Antenna, 24 dBi GainMaximum
60 GHz

24 dBiLinear
9.7°
11.0°-13 dB-36 dB
23 dB+25 °C-40 °C+85 °C

Mechanical Specifications:
ItemAntenna PortFlange TypeSize
Material
FinishWeight
OutlineSpecification
WR-19 WaveguideUG-383/U-M Anti-Cocking Flange
3.90” (L) X 1.75” (W) X 1.42” (H)BrassGold Plated
2.78 Oz
AZ-U24-A
www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com

Copyright © 2022 by Eravant

Page | 1

SAZ-2410-19-S1

Rev. 1.5Simulated Antenna Patterns @ 50 GHz
3020H-Plane
E-PlaneGain (dBi)

100-10-20-30-40-180-150-120-90-60-30

0306090120150180

Boresight Angle (Degrees)

SAGE Millimeter, Inc.

WR-19 Standard Gain Horn Antenna, 24 dBi Gain

Simulated Gain vs. Frequency
26

Gain (dBi)2524232221

204041424344454647

48

49

50

51

5253

54

55

56

57

58

5960

Frequency Range (GHz)www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com
Copyright © 2022 by Eravant

Page | 2

SAZ-2410-19-S1

Rev. 1.5

Simulated Gain vs. Frequency in Tabular Format
Frequency (GHz)404142434445
4647484950

Gain (dBi)
22.8
23.1
23.1
23.3
23.4
23.6
23.7
23.8
23.9
24.0
24.1

Frequency (GHz)
5152
53
54
55
56
57
58
5960

Gain (dBi)
24.3
24.2
24.4
24.5
24.5
24.7
24.7
24.7
24.9
24.8

Simulated Half Power Beamwidth (H-Plane) vs. Frequency in Tabular Format
Frequency (GHz)404142434445
4647
484950Beamwidth (Degrees)
12.412.3
12.1
11.9
11.8
11.6
11.4
11.3
11.1
11.1
10.9

Frequency (GHz)
51
52
53
54
55
56
57
58
5960

Beamwidth (Degrees)
10.9
10.7
10.6
10.5
10.4
10.2
10.2
10.1
9.9
10.1

SAGE Millimeter, Inc.

WR-19 Standard Gain Horn Antenna, 24 dBi Gain

Simulated Half Power Beamwidth (E-Plane) vs. Frequency in Tabular Format
Frequency (GHz)
40
41
42
43
44
45
46
47
48
49
50

Beamwidth (Degrees)
11.3
11.1
11.0
10.7
10.6
10.5
10.3
10.2
9.9
9.8
9.4

Frequency (GHz)
51
52
53
54
55
56
57
585960Beamwidth (Degrees)
9.0
8.9
8.4
8.3
8.0
7.7
7.8
7.3
7.3
7.2

www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com
Copyright © 2022 by Eravant

Page | 3

SAZ-2410-19-S1

Rev. 1.5

Mechanical Outline: (Unless otherwise specified, all dimensions are in inches [millimeters])

Note:
••

•
•

All data presented is simulated by a full EM simulator. Eravant recommends using simulated
data over measured for standard gain horn antenna for accuracy. See Blog here for furtherinformation.
The antenna electrical performance is guaranteed through accurate mechanical tolerance
control. Each antenna is examined by CMM (coordinate Measuring Machine) inspection andmeasurement process.
A calibration certificate can be issued with a fee under part number FTA-0150-S1-SAZ.
Eravant reserves the right to change the information presented without notice.

SAGE Millimeter, Inc.

WR-19 Standard Gain Horn Antenna, 24 dBi Gain

Caution:•Any foreign objects in the antenna will cause performance degradation and possible device
damage.

www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com
Copyright © 2022 by Eravant

Page | 4