Rline®
Dispenser
Modulefor Your Liquid
Handling Automate

Rline® Dispenser Module
Proven Technology for Your Automate
The Sartorius Rline® Robotic Dispenser Module brings theproven technology of Sartorius electronic pipettes into thefield of laboratory automation. The dispensing head module
is designed to be used as the liquid handling component of
any automated system and has a successful track record invarious customer applications stretching back several years.

-Save Space	The compact size and light weightof the Rline® module ensures aneasy fit with any instrument design.
Save Time	The dispensing head is easy toconnect, ensuring quick integration
and rapid time to market.

Achieve Reliable Results	The innovative, proven features
enable high accuracy and precision.

Features
The unique and innovative features of the Rline® Robotic
Dispenser Module ensure the accuracy and performanceof sample processors and automated instrumentation:-	Outstanding accuracy and reliability with the enhanced
DC-motor concept, which includes an electronic brakesystem and optical sensor.	The combination of Liquid Level Sensor with conductive
Robotic Tips enables the detection of the liquid surface
in the target vessel. This prevents erroneous attempts toaspirate liquid before the liquid surface has been reached.
Liquid level sensing feature is available with Rline 200and Rline 1,000 models and requires the use of conductive tips.

Rline® Dispenser Module
Suitable for Various Applications---------

Typical Applications

Technical Features	Laboratory automation for research and clinicalapplications
Laboratory robotics and automated workflows
Industrial applications and production systems	Single-channel models cover a volume rangeof 5 to 5,000 µl	Electronic tip ejection	Communication protocol RS232, 9.6 – 115.2 kbps	RS485 communication available on request	Low tip-loading forces, 12 – 45 N	Operating voltage: 9 – 12 VDC	Current consumption:idle <100 mA, average < 500 mA, peak 2 A	Power and communication connector: Tyco 609-1427	Conforms to ISO 8655 requirements	ISO 17025 accredited calibration certificates available
on request	Customizable as an OEM product on request

The Rline® can be used as an off-the-shelf module
or tailored to fit a specific application.

Detailed technical manual available on request.

Technical Features
Rline®
OrderCodeChannels

VolumeRange(µl)

71099315 – 200

7109957109691150 – 1,000

100 – 5,000

LiquidLevel
Sensing*-Increment(µl)

Test
Mode**

Test
Volume(µl)

Systematic
Error Limit(%)***

Systematic
Error Limit(µl)***

Systematic
Error Limit(%)***

RandomError Limit(µl)***

0.5

P200

1.0

2.00

0.3

0.50

2.510P

100

0.8

0.80

0.25

0.25

P20

2.00.40

1.00.20P5

4.00.20

2.00.10D10 × 20

2.00.40

2.00.40

P

1,000

0.45

4.50.15

1.5P500

0.6

3.00.2

1.0P100

2.0

2.0

0.5

0.5P504.0

2.0

1.00.5d10 × 100

2.5

2.5

2.0

2.0P

5,000

0.5

250.15

7.5

P

2,500

0.7

17.5

0.25P500

1.6

80.4

2P100

8.082.02d10 × 500

2.4122.4

12* Note. Requires conductive robotic tips** Note. P = pipetting, D = multi-dispensing*** Note. The listed systematic and random error values can be achieved only under strictly controlled conditions during type tests per ISO 8655.
Pipettes are tested at factory default speed settings. Due to the continuous product development by Sartorius, the systematic and random error
values are subject to change without prior notice.

Development Kits

Pipette Tips

The unique and innovative features of the Rline® Robotic
Dispenser Module ensure the accuracy and performance
To help you to get started with Rline, we also offer
development kits for all models. Development kits include:
1 Rline of you choice, complete technical documentation,
power supply and data cable, PC software, and a tray of tips.

Sartorius offers Robotic Tips, wich are specially designed
for use with the Rline® Liquid Level Sensing model.
Conductive Robotic Tips enable the detection of the liquidsurface in vessels or microplates. Please contact us for theconductive tips.

Development kits can be ordered using the following codes:
LH-710993KIT for Rline 5 – 200 µl
LH-710995KIT for Rline 50 – 1,000 µl
LH-710969KIT for Rline 100 – 5,000 µl

Tips
Order CodeVolume Range (µl)

7902005 – 200790201F

5 – 200780304

100 – 5,000

Filtered-Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0
 or further contacts, visitFwww.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 10 | 2020

Conductive(for level sensing)

Sterile-USA
Sartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906

Quantity (tips)

Compatibility

10 × 96

Rline 5 – 200 µl

10 × 96

Rline 5 – 200 µl

50

Rline 100 – 5,000 µl

Finland & Baltics
Sartorius Biohit Liquid Handling Oy
Laippatie 100880 Helsinki
Phone +358 9 755 951