User Manual

PRECISION MEASURING TOOLS

Model No. D-***HL

LEFT HAND DIGITAL VERNIER CALIPERS
Thank you for purchasing Niigata Seiki Digital Calipers.
This is a precision measuring tool for making Inside, Outside, Step Height, and Depth measurements.
●For safe and proper use, please read this instruction manual before use and follow the procedures described.
Please keep manual where it is accessible to user for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact dealer or Niigata Seiki.

■PART IDENTIFICATION AND FUNCTION※Model D-150HL shown.

⑨ Lock Screw⑩ LCD Display⑧ Scale⑥ Depth Bar⑤Measuring surfaces forinside measurements④ Inside Jaws

② Outside Jaws

⑪ Battery Cover⑬ [ON/OFF] Button①Measuring surfaces foroutside measurements①Measuring Surfaces for outside measurements … Object to be measured is positioned between these surfaces.
②Outside Jaws ……………………………………… For measuring outside dimensions.
③Measuring surfaces for step measurements …… Measures height of steps.
④Inside Jaws………………………………………… For measuring inside dimensions.
⑤Measuring surfaces for inside measurements …… Position surfaces inside object to be measured.
⑥Depth Bar ………………………………………… For measuring depth.
⑦Measuring surfaces for depth measurements …… Depth is measured between these two surfaces.
⑧Scale ……………………………………………… Supports LCD Display Unit.
⑨Lock Screw ……………………………………… Tighten to Prevent the Slide and Display from moving.
⑩LCD Display ……………………………………… Displays digital measurement.
⑪Battery Cover …………………………………… Open to access battery.
⑫[ZERO] Button …………………………………… When pressed, Display will read "0.00".
⑬[ON/OFF] Button ………………………………… For turning power ON /OFF.

To prevent harm to yourself or others, and to prevent damage toproperty, always follow the procedures marked by the following
symbols.

WARNINGUse only for measuring.
・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.
Handle with Care.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・Do not scratch instrument, for example by writing ID
number.
Avoid using in high electric fields, such as nearfluorescent lights or switching power supplies.
・Incorrect value may be displayed on LCD.

Contamination of surfaces may cause measurement
error.

A temperature difference between the object to bemeasured and the caliper may cause measurement
error. Allow enough time for the temperatures toequilibrate.

INSERTING・REPLACING BATTERY① Slide cover in direction of arrow.
② Insert battery with (+) side outward.
③ Replace battery cover.

Denotes a prohibition You MUST NOT do

Use only SR44 type battery.

BatteryCover

RemoveDenotes a requirement You MUST do

Use and store only in environment meeting thefollowing conditions:●Dry location protected from rain, water, and oils.
●Location that is not excessively hot.
●Location protected from use by children andunauthorized people.
・Use in places contrary to the above may cause
damage to the product resulting in accident or injury.
Jaw Tips Are Sharp̶Handle With Care.
・Careless handling may cause injury.
Do not disassemble or modify.
・Disassembly may damage product or cause accident.
For repair, please contact distributor .
Do not write on instrument with electric engraver.
・Do not engrave instrument body or jaw, such as an ID
number with engraver or electric pen.

HOW TO USE■Power ON/OFF①To turn on, press the [ON/OFF ] Button.
②To turn off, press the [ON/OFF ] Button.
※Power will automatically turn OFF after about 20 minutes with noactivity. Press the [ON/OFF ] Button or move the Slide to turn ON.

■Absolute Measurement Mode

①Loosen the lock screw, and turn on power.
②With jaws closed, press the [ZERO] Button.
③When “0.00mm” is displayed on LCD, Calipers is ready tostart measurement.

■Comparisons Measure Mode

①Loosen the lock screw, and turn on power.
②With jaws open to desired reference position (any pointis OK,) press [ZERO] Button.
③When “0.00mm” is displayed on LCD, Calipers is readyto start measurement.

AFTER USE CARE・STORAGE●After use, wipe off any dirt or oil frommeasuring surfaces, sliding surfaces, andexterior using a dry cloth and perform rustprevention treatment.
For rust prevention, lightly apply anti-rust oil, or place inrust preventive bag.

●Store with Jaws slightly open.

If Jaws are closed, expansion due to temperature changes
may cause excessive force which will cause inaccuracy.

●Store in a cool, dry, and dark location in theprovided case.
Keep out of direct sunlight and moisture, and pleasekeep secure from unauthorized personnel.

●Remove battery if Caliper will not be used for along period of time.
Batteries may leak if left in unused instrument for anextended period.

TROUBLESHOOTING

Please Observe

Indicates risk of personal injury or property damage if not followed.

Read the manual and follow the directions.
・Use of product other than as described in this manual
may cause accident.

●Wipe off rust proofing and dirt from measuring
surfaces and slide.

Notice⑫ [ZERO] Button

SAFETY PRECAUTIONSBefore using the calipers, please follow the steps below:●Allow the calipers and object to be measured
time to reach the same temperature.

③Measuring surfaces
for step measurements⑦Measuring surfaces
for depth measurementsPREPARATION FOR USE

Q. Nothing is displayed on LCD.
A. The battery may be running low, please replacebattery with a new one.
Q. With power on, the number on the LCD does notchange as Display Unit is moved.
A. Please reset by removing battery for about 30sec.,
and then replacing.
Q. Nothing happens when buttons are pressed.
A. Make sure battery is properly inserted.
Q. The correct value is not displayed.
A. High humidity may cause dampness. Please dry off
the instrument and use only in dry location.

SERVICE●If Instrument is not working properly, or if you have anyquestions, please contact distributor.
●Please note, manufacturer is unable to respond toinquires or provide service directly. Please contact
distributor .

PRODUCT SPECIFICATIONS
Model No.

Measuring
Range

Maximum ※permissible error

D-150HL
D-200HL
D-300HL

0 -150 mm
0 -200 mm
0 -300 mm±0.03 mm±0.04 mm

Weight150g
190g
250g

※Not including quantization error (±1 count)●Resolution : 0.01mm●Display : 5 Digit LCD●Auto Off : approx.20 min.
●Maximum response speed : Slide to Right .....1.5 m/sSlide to Left .......0.75 m/s※Left/Right as shown in diagram at top of manual.

●Operating Temperature Range :0°
C ~40°C●Storage Temperature Range : −10°
C ~ 60°C●Power : (1x) SR44 Silver oxide Battery●Battery Life : approx. 1 years in normal use●Accessory : SR44 Silver oxide Battery (for test)

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書 品番

精密測定工具左勝手デジタルノギス

D-***HL

ご使用前に、下記の準備を行ってください。

●測定面、摺動面、外観などの汚れや防錆油などを拭き取る

この度は「左勝手デジタルノギス」をお買上げいただきありがとうございます。
この商品は、測定物の外側・内側・段差・深さ測定が可能な精密測定工具です。

そのまま測定を行うと誤差の原因となります。

●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称と機能 ※イラストはD-150HLです。
⑥ デプスバー⑤内側用測定面⑨ 止めねじ⑩ LCD表示部⑧ 本尺④内側用ジョウ
③段差測定面②外側用ジョウ

⑪電池蓋⑦ 深さ測定面

使用前の準備⑫ZEROスイッチ●本体と被測定物の温度慣らしを十分に行う測定器本体と被測定物の間に温度差があると、誤差の原因となります。同じ温度になるよう、温度慣らしは十分に行ってください。

電池のセット・交換①電池蓋を矢印の向きに開けます。
②電池を、必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。
ご注意 電池は必ずSR44をお使いください。
電池蓋⑬ ON/OFFスイッチ①外側用測定面①外側用測定面 …… 被測定物をこの面で挟んで測定します。

⑧本尺 ……………… デジタル液晶部のスライドを保持します。

③段差測定面 ……… 段差面に当てて長さを測定します。

⑩LCD表示部 ……… 測定数値をデジタル表示します。

②外側用ジョウ …… 被測定物の外側を測ります。
④内側用ジョウ …… 被測定物の内側を測ります。

開ける⑦深さ測定面 ……… 測定面に当てて深さを測定します。

安全上のご注意
注意

必ず実行していただく
内容『強制事項』です。

お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

取扱説明書をよく読み、指示に従う。

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

測定作業にのみ使用する。

下記の条件を満たす環境で使用・保管する。
●雨や水、油などがかからない、乾燥した場所●高温にならない場所●子どもや、使用者以外が近付かない場所

・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。

・上記に反する場所での使用は製品の破損、事故やけがの原因となります。

・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・商品に傷をつけないでください。

・けがの原因となります。

大切に取り扱う。

インバータ式蛍光灯などの電界が発生する場所での使用は避ける。

・ＬＣＤ表示部の数値が誤作動を起こす原因となります。

ジョウ先端は尖っているので、扱いに注意する。
分解・改造を行わない。

・製品の破損や事故の原因となります。修理はお買い上げの販売店、または発売元までご連絡ください。

電気ペンなどで書き込まない。

・電気ペンなどで本尺やジョウなどに、番号などの書き込みをしないでください。

①止めねじをゆるめ、電源を入れてください。
②ノギスのジョウを閉じた状態で "ZERO"スイッチを押してください。
③LCD表示部に"0.00mm"が表示されると計測開始ＯＫです。

■比較測定モード①止めねじをゆるめ、電源を入れてください。
②ノギスのジョウが開いた状態（どのポイントでもＯＫ）の時に"ZERO"スイッチを押してください。
③LCD表示部に"0.00mm"が表示されると計測開始ＯＫです。

使用後のお手入れ・保管方法●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う
防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

●ジョウを少し開いた状態で保管する●長期間使用しない場合は電池を製品から取り外した状態で保管してください⑬ON/OFFスイッチ … 電源のON/OFFを行います。

してはいけない内容『禁止事項』です。

■絶対測定モード

直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

⑪電池蓋 …………… 開けて電池をセットします。

長期間使用しないで放置すると、電池の液漏れの原因となります。

トラブルと対応

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを右記のように説明しています。

①電源をONにするには、”ON/OFF”スイッチを押してください。
②電源をOFFにするには、”ON/OFF”スイッチを押してください。
※約20分間無操作状態が続くと、自動的に電源がOFFになります。”ON/OFF”スイッチを押すか、ジョウをスライドさせると電源がONになります。

●付属のケースに入れ、乾燥した冷暗所に保管する⑨止めねじ ………… デジタル液晶部のスライドを固定します。

必ずお守りください■電源のON/OFF

ジョウを閉じた状態で保管すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

⑤内側用測定面 …… 被測定物の内側にこの面を当てて測定します。 ⑫ZEROスイッチ …… 押すと画面数値が"0.00"と表示されます。
⑥デプスバー ……… 深さを測定するバーです。

使用方法

Ｑ．画面に何も表示されなかったり、数字が点滅している。
Ａ．電池の残量が少なくなっている可能性があります。新しい電池と交換してください。

Ｑ．電源ONの状態でスライドさせてもLCD表示の数字が変わらない。
Ａ．一度電池を取り外し、30秒後に電池をセットし直してください。
Ｑ．スイッチを押しても何も反応しない。
Ａ．電池が正しくセットされているか確認してください。

Ｑ．正しい数値が表示されない。
Ａ．使用環境の湿度が高い可能性があります。本器を乾燥させ、湿度が低い環境下でご使用ください。

修理の際は●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、当社までご連絡ください。

●お問い合わせや、ご連絡が無いまま直接当社に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

製品仕様※量子化誤差（±1カウント）を含まず

品 番

測定範囲（mm）

最大許容誤差 ※（mm）

質量（g）

D-150HL
D-200HL
D-300HL

0 -150
0 -2000 -300±0.03150190250±0.04●最 小 読 取 値：0.01mm●表示：LCD 5桁表示
●オ ー ト オ フ：約20分●最大応答速度：右方向へのスライド…1.5m/s
左方向へのスライド…0.75m/s※右方向、左方向は各部の名称の図を正面にした場合●使用温度範囲：0℃〜40℃●保存温度範囲： −10℃〜60℃
源：酸化銀電池 SR44×1●電●電 池 寿 命：通常使用状態で約1年●付属品：酸化銀電池 SR44（テスト用）〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp2211

I92-K4