800 Series
DC Hipot Testers - 15kV, 80kV, 120kV & 170kV The 800 Series DC Hipot Testers is an
economical solution to DC field testing of cables,terminations, motors, generators and other electricalapparatus. All models are self-contained in a one or
two-piece, rugged and durable enclosure. Each iscomplete from input line cord to high voltage output
cable. This product range includes models in 15kV,80kV, 120kV and 170kV DC.

FEATURES

The 800 Series features accurate kilovoltmeterreadings regardless of load current. Voltage
measurements are taken directly at the output of thehigh voltage transformer, while current is measured
at the return leg to ensure the highest accuracy.
Safe discharging of both the test object and internaltransformer occur whenever high voltage is switched
off. Output power is created through a full-wave,voltage doubling, silicon rectifier circuit.

Environmental friendly FR3™ transformer oil

Rugged and portable construction

Shielded output cableFull-wave voltage doubling rectifier

Zero start interlock and guard circuitInternal discharge solenoid

Meter accuracy ± 2% full scale

5-10 mA current ratingExternal interlock provisions



Three range voltage meter

Four range current meter

No internal leakage at full load

Instantaneous overload relay

Surge-limiting resistors in High Voltage
outputBENEFITS
Ideal for field testing applications.
Automatic grounding of power supply and test object
when high voltage is turned OFF.
Minimal setup time and simple control panel.
Accurate current measurement and guard circuitdesigned to eliminate stray leakage currents.

APPLICATIONS




One-Piece Design for 15kV & 80kV models
Controller for 120kV & 170kV models

1/2Cable
Transformers
Electrical Switchgear
MotorsGenerators
Other Electrical Apparatus

TECHNICAL SPECIFICATIONS
Model Number
System Output (V)System Output (A)Polarity
Metering Accuracy
RippleDimensions(W x H x D)

815PL-*
0 - 15 kV DC
5mA

880PL-10mA-*
8120-5PL-*
8170-5PL-*
0 - 80 kV DC
0 - 120kV DC
0 - 170kV DC
10mA
5mA
5mA
Negative Output, Positive Ground±2% of full scale
0.5% per mA with rated resistanceLess than 2.5%ControllerHigh Voltage
SectionControllerWeightHigh
Voltage
SectionInput Frequency
Input Voltage

16.5 x 9 x 18 in (42 x 23 x 46 cm)
In Controller

12 x 10 x 19 in(30 x 25 x 48 cm)

12 x 10 x 30 in(30 x 25 x 75 cm)

Net

49 lbs (22 kg)

78 lbs (35 kg)

25 lbs (11 kg)

Shipping

64 lbs (29 kg)

90 lbs (41 kg)

35 lbs (16 kg)

In Controller

In Controller

Net
Shipping

Included Accessories

102 lbs (46 kg)

145 lbs (66 kg)

122 lbs (55 kg)

165 lbs (75 kg)

50/60 Hz
* In the model number, designate 'A' for 120V input or 'B' for 230V input
Input Cord, 6 ft (1.8 m)
Return Cable, 25 ft (7.6 m)
High Voltage Output Cable,
shielded RG8/U with alligator clip
Interlock Plugand rubber insulated boot, 25 ft(7.6 m)
Calibration Certificate
User’s Manual

SYSTEM CONTROLSHigh Voltage Section
for 120kV & 170kV units(pictured right)

OPTIONAL EQUIPMENT & ACCESSORIESPart Number
HHDA13-280
SPK1-815PL
SKP1-880PL-10MA
SPK1-8120-5PL
SPK1-8170-5PL
EXT-WARN-1

Description
Grounding Stick, 120kV Max Voltage
Spare Parts Kit for 815PL
Spare Parts Kit for 880PL-10mA
Spare Parts Kit for 8120-5PL
Spare Parts Kit for 8170-5PL
One year extended warranty

2/2

HHDA13-280 Grounding Stick(pictured above)