DIGITAL PROTRACTOR INSTRUCTION MANUAL
Model No.: DP- 90G EX), percent●Angle can be measured as: degree (°slope (%), or gradient (mm/M).
●Display will automatically flip if gauge is upside
down for easy reading of overhead
measurements.
,●Audible beep feature when measurement is: 0°90°, 0%, or 0 mm/M.
●Back-light feature for reading in low light areas.
Press the [ 照明 ] Button (Light) and Back-lightwill turn on for approx. 1 min.
●Hold Button to allow reading when display isinaccessible. Press [ 表示保持 ] Button (Hold)and display reading will hold.
●V-Groove base for measuring cylindricalsurfaces. Also, Base has four magnets forvertical or overhead measurements on steel.

APPLICATIONS●Piping and equipment construction and installation.
●Roofing and exterior construction.
●Tile cutting and setting.
●Steel and civil engineering measurement andconstruction.

SPECIFICATIONS＊Specifications subject to change
without notice.

●Model : DP-90G EX●Measuring Range : ±90.0°●Resolution : Angle Display = 0.05°
Ercent Display = 0.1%
Gradient Display = 1mm/M●Accuracy : 0.0°
〜 90.0°=±0.3°●Display Stabilization Time : approx. 10 sec.
●Operating Temperature : 0 〜50°C●Batteries : 9V Battery●Dimension : L173×W47×H68mm (excluding protrusions)●Weight : 550g
●Accessories : Soft case, Manual, 9V Battery (for test)

OPERATION
ACCURACY CONFIRMATION〔Horizontal Accuracy Check〕(Figure:1)
3. Place gauge on a horizontal surface and turn on the power. After
display has stabilized, rotate 180°at the same position. When display
is stabilized, the reading should be same as before to confirm
accuracy.
(If readings are withing normal range of instrument error of ±0. 3°)(Figure:1)〔Vertical Accuracy Check〕(Figure:2)
4. Place gauge on a vertical surface with display side up, and turn on
the power. After display has stabilized, rotate gauge 180°in
horizontal direction at the same position, keeping the display on theupper portion. When display is stabilized, the reading should besame as before to confirm accuracy.
(If readings are withing normal range of instrument error of ±0. 3°)＊If accuracy checks, gauge is ready to take measurements.

SETTING ZERO AT ARBITRARY ANGLE(Figure:2)＊Zero Point can be set at any angle of inclination.
1. At desired angle, wait for display to stabilize and press the [ 表示保持]Button (Display Hold).
2. When the display blinks, press the [校正/ゼロ] Button (Calibration/Zero).
This angle is now set as zero.
3. Press [表示保持] Button (Display Hold) again to cancel this setting.

CALIBRATION

Rotate to remove cover whenreplacing Battery.

DIGITAL DISPLAY①Arrow Display (Ex:indicates angle)①②①●Indicates direction to adjust tilt for level. Arrows display forangles between 0.05°〜44.9°
to help in adjusting forhorizontal. (In Figure, arrows are shown for example with rightside of gauge elevated.)●Arrows reverse at 45.1°for use as reference in verticalmeasurements. (Gauge can be used in horizontal or verticaldirection.), 90.00°and display as●Arrows shrink and go out at 0.00°④③④straight bar at 45.0°②Battery Indicator
Automatically displays when battery is low.
③Buzzer Icon
Press [ ブザー ] Button (Buzzer) to turn on icon and buzzer will beep at 0°, 90°, 0% and 0mm/M. Pressagain to disable buzzer.
④°, ％, mm/M Icon), percent slope (%), or gradient (mm/M).
Indicates display angle is in degrees (°⑤Display Switching Button),
Changes display angle - degrees (°percent slope (%), or gradient(mm/M).
⑥ON/OFF Button (Power Button)
Press to turn on power, press again
to turn off.
Power will turn off automatically ifgauge is idle for about 6 min.
⑦Buzzer Button⑤⑥ ⑦ ⑧ ⑨ ⑩
Press once and buzzer will beep at0°, 90°, 0%, 0mm/M. Press again to disable buzzer.
⑧Calibration / Zero Button
For use when calibrating horizontal and vertical setting, and for setting an angle as Zero-Point. (Pleaserefer to: “Setting Zero,” and “Calibration” sections).
⑨Light Button
Press once to turn on back-light, press again to turn off.
Light will turn off automatically in about 1 min.
⑩Display Hold Button
Press once to hold value on display. Press again to release value and resume measurements.

beep

1. Only press [ 校正 / ゼロ ] Button (Calibration/Zero) during calibration or zero setting. Pressing at any othertime will cause inaccurate measurements.
2. When battery is replaced, power will automatically turn on. This is normal operation; to turn off press the[ON/OFF] Button.
3. Do not drop or subject to shock as it may cause poor accuracy.
4. The bottom of the instrument contains magnets, please keep from keep away from sensitive items suchas computers, watches, compass, etc. as it may cause a malfunction.
5. Do not disassemble instrument under any circumstances.

DISPLAY ORIENTATION＊For angle about 22°beyond 90°, Display will automatically flip to make it easier to read. (Display value of)about 68°
Please use care when taking measurement to avoid misreading display. (Figure:11)（Figure:3）〔Angle Examples〕

90° 22°

beep

approx.68°
Flip（Figure:4）
beep

CAL３

（Figure:5）
beep

CAL４

（Figure:11）0.00°
・After the display flips, to return to original
direction, move back toward starting angle. For
angle about 22°beyond 90°in the otherdirection, display will return to originalorientation.
(Display value around 68°) (Figure:12)22°

6 8°10. Hold the Base of the Gauge on a flat surface which is roughly
vertical with the Display side up.
11. After about 15 seconds, gauge will <beep>, and display will
flash to “CAL5 ”. (Figure:7 )Display will then change to “UP” ←→”-6-” text.
12. Rotate instrument 180°horizontally, and hold to same location
on surface.
13. Gauge will <beep>, and display will flash to “CAL6”. (Figure:8)
Display will then change to “UP” ←→”-7-” text.
14. Next step is to repeat for gauge with display in the lowerposition.
15. Gauge will <beep>, and display will flash to “CAL7”. (Figure:9 )Display will then change to “UP” ←→”-8-” text.
16. Rotate instrument 180°horizontally, and hold to same location
on surface.
17. Gauge will <beep>, and display will flash to “CAL8”. (Figure:10)＊Calibration is now complete.

Battery Cover Screw

Base Reference Surface(With 4 Magnets)

6 8°〔Vertical Calibration〕

V Groove

CAUTION＊Calibration is required if accuracy check error is greater than ±0. 4°, or under certain conditions:●If instrument is subject shock or vibration, or large changes in temperature.
●If battery is removed, as when gauge is not used for a long time.
＊Full calibration procedure of steps 1 to 17 below must be performed. Follow the figures, and us the display“UP” text as a guide.
1. Place the Base of the Gauge on a clean, flat surface which isroughly horizontal and press the [ON/OFF] Button to turn on.
2. After about 15 seconds, press the [校正/ゼロ] Button(Calibration/Zero) and hold for two seconds.
“CAL” will flash on display, and then display will change to“UP” ←→”-1-” text.
3. After about 15 seconds, gauge will <beep>, and display willflash “CAL1”. (Figure:3 )Display will then change to “UP” ←→”-2-” text.
4. Rotate instrument 180°horizontally, in same location onsurface.
5. Gauge will <beep>, and display will flash “CAL2 ”. (Figure:4 )Display will then change to “UP” ←→”-3-” text.
6. Next step is to repeat for top surface of gauge.
7. Gauge will <beep>, and display will flash “CAL3 ”. (Figure:5 )Display will then change to “UP” ←→”-4-” text.
8. Rotate instrument 180°horizontally, in same location on surface.
9. Gauge will <beep>, and display will flash “CAL 4”. (Figure:6 )Display will then change to “UP” ←→”-5-” text.
＊Horizontal calibration is now complete, Continue on to perform
vertical calibration.

Tightenｍｍ/Ｍ

OPERATION BUTTON

1. Make sure Base Reference Surface is clean and free of adheredcontamination or dents.
2. Press the [ON/OFF] Button to turn on, and check icon to confirm
proper units ( °, %, mm/M).

〔Horizontal Calibration〕

Loosen
単位 切替°／ ％ ／6 8°FEATURES〔Rear〕

Digital Display

6 8°

DESCRIPTION

MAIN BODY〔Front〕

Thank you for purchasing the Digital Protractor.
Please read this manual thoroughly before use for proper operation.

This angle gauge uses a liquid sensor to measure
the angle of inclination of the instrument.
The change in liquid level is converted to acorresponding electrical signal and displayed onthe LCD.

PART IDENTIFICATION and FUNCTION

Flip

・If position does not exceed 90°, the display will
not flip.

（Figure:12）

CONVERSION TABLE（Figure:6）
beepbeepa. Gradient
Gradient

1/200

1/150

1/1151/100

1/80

1/50

1/40

1/30

1/20

1/10

Percent

0.5％

0.7％

0.9％

1.0％

1.3％

2.0％

2.5％

3.3％

5.0％

10.0％

Degrees

0.3°

0.4°

0.5°

0.6°

0.7°

1.1°

1.4°

1.9°

2.9°

5.7°

Degrees

0.0°

10.0°

20.0°

30.0°

45.0°

60.0°

70.0°

80.0°

90.0°

Percent

0.0％

17.6％

36.4％

57.7％

100％

57.7％

36.4％

17.6％

0.0％

b. Angle〔Angle Diagram〕（Figure:7）（Figure:8）（0％）（17.6％）
90° 80°〔Gradient Diagram〕
45°（100％）（10％＝0.1）

beepCAL７1/10beep

CAL８

130°
（57.7％）
20°（36.4％）

1010°（17.6％）
0°（0％）（Figure:9）（Figure:10）＊ When displaying percent slope, gauge will display angle(0% to 100%),
referenced to horizontal from 0°to 45°(100% to 0%).
and referenced to vertical for 45.1°
to 90°

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

デジタル角度計レベルニック

取扱説 明書

品番：DP-90GEXこの度はデジタル角度計レベルニックをお買上げいただきありがとうございます。
ご使用前に、本書をよくお読みいただき正しくお使いくださいますようお願い申し上げます。

概 要

本器は液体センサーを使用した角度計であり、傾斜角に応じた液体レベルの変化を電気信号に変換してデジタル表示をしています。

特 長●角度（°）、割合（％）、傾斜（mm/M）を測定できます。
●本器を逆さまに使用した場合、デジタル表示も自動的に逆転しますので天井測定の場合に便利です。

、90°、0 ％、0 mm/M でブザー音を出すことがで
●0°
きます。

［照明］ボタンを押すとバックライトが点灯し、暗●い場所でも読み取りできます。
（約１分で自動消灯）●目の位置より高い場所での測定でも［表示保持］ボタンにより手元に持ってきて読み取りできます。

●ベースはＶ溝付きですので円筒面上の測定もできます。また、基準面は磁石４個付きで、鉄骨などの垂直面や天井の測定が楽に行えます。

各部の名称と機能
本 体

デジタル表示部

〔正 面〕

〔背 面〕

用 途

●配管、設備工事の測定と設定●屋根、外壁工事の測定と設定●左官、タイル工事の測定と設定●鉄骨、土木工事の測定と設定

仕 様

ゆるむ

Ｖ溝

電池蓋ツマミ

ベース基準面

電池を交換する時に回して電池蓋をはずしてください。

（磁石４個付き）＊予告なしに仕様変更することがあります。

式：DP-90G EX●型●測 定 範 囲：±90.0°●最 小 読 取 値：角度表示＝0.05°
割合表示＝0.1％
傾斜表示＝ 1 mm/M
●読 取 精 度：0.0°〜 90.0°＝±0.3°●表示安定時間：約 10 秒
●使 用 温 度：0 〜 50℃●使 用 乾 電 池：9V 角形乾電池●外 形 寸 法：L173 × W47× H68mm（突起物を除く）●重量：550g
品：ソフトケース・取扱説明書●付 属
9V 角形乾電池（テスト用）

しまる

デジタル表示部①矢印表示 （例、角度の場合）●傾斜している場合の調整方向を示しています。（図は本器が右上がりになった例です。）0.05°〜 44.9°までは水平方向に調整するように矢印がでます。
以上になると矢印が反転し垂直方向に調整するように矢印●45.1°がでます。（本器は水平と垂直に基準をもっているため）、90.00°では矢印が消え、45.0°
はバーのみの表示になります。
●0.00°①②電池表示電池の容量が少なくなると自動的に表示します。

②④

①③④

③ブザー表示〔ブザー〕ボタンを押すとブザー表示がつき、0°、90°、0％、0mm/M でブザー音が鳴ります。 再度ボタンを押すとブザー表示は消え、ブザー音は鳴りません。
④ °、％、mm/M 表示
現在表示している数値が角度表示（°）か、割合表示（％）か、傾斜表示（mm/M）かを表しています。

使用方法

操 作 部

⑤表示切替ボタン
角度（°）、割 合（％）、傾斜（mm/M）を切り替えます。

精度確認

１．ベース基準面に、ゴミ等の付着や打痕等による突起がないことを確認してください。
２．電源ボタンを押し、デジタル表示をしているか、または °、％、mm/Mのいずれか表示していることを確認してください。

&2)':⑥ＯＮ／ＯＦＦボタン（電源ボタン）
１回押すと電源が入り、再度押すと電源は切れます。
また、電源を入れたままの状態にしておきますと約 6 分後自動的に切れます。

〔水平精度確認〕（図１）
３．
水平の出ている面に本器を置いて電源を入れます。表示値が安定した後、同一面上で１８０°回転します。安定した後の表示値が変わらないことで精度確認をします。
（また、表示値の差がある場合でも ±０．
３°以内であれば正常です。）（図１）〔垂直精度確認〕（図２）
４．
垂直の出ている面に本器のデジタル表示部が上になるようにベース面をあて、電源を入れます。デジタル部を上のまま水平方向に１８０°回転し、本器上面を同一面上にあてます。安定した後の表示値が変わらないことで精度確認をします。
（また、表示値の差がある場合でも ±０．
３°以内であれば正常です。）（図２）＊任意の角度をゼロとして測定することができます。
１．任意の角度で表示が安定するのを待ち、［表示保持］ボタンを押します。
２．表示が点滅したら、［校正／ゼロ］ボタンを押します。これで任意ゼロが設定されました。
３．本設定を解除する場合は、もう一度［表示保持］ボタンを押します。

１．校正や任意ゼロ設定の場合以外に、［校正／ゼロ］
ボタンを押さないでください。精度が狂う原因となります。

２．電池を交換した場合、電源が自動的に入りますが異常ではありません。そのまま［ＯＮ／ＯＦＦ］
ボタンを押すと電源は切れます。
３．落下等、強い衝撃を与えないでください。精度不良の原因になります。

４．本体底面のマグネットを、コンピュータ・時計・精密機器・方向コンパスなどに近づけないでください。故障の原因になります。

＊次のような時には精度確認をし、±０．
４°以上の誤差を生じた場合は校正を行う必要があります。
●気温の激しい変化、本器が強い衝撃や振動を受けた場合●乾電池を抜いて長時間使用しなかった場合など＊校正は必ず下記手順１．〜17．を通して行ってください。

＊本器は90°を超え、さらに約22°傾けると表示を読みやすくするために、表示値が自動的に逆転します。（表示値は約68°）
測定の際は間違えないようにしてください。（図１１）

ピッ

CAL３

＊これで水平の校正が完了しました。引き続き垂直の校正を行います。

（図５）
ピッ

CAL４

（図６）

ピッ

15．
「ピッ」と音が鳴り、［ＣＡＬ７］が表示されます。
（図９）
その後［ＵＰ］←→［-８-］が切り替り表示されます。

・一旦逆転した表示値を元に戻す場合はそのまま元の角度の方向に傾きを変えると、90°を超え、さらに約22°の所で元にもどります。（表示値は約 68°）
（図１２）

6 8°
22°

・水平状態より右上がり、左上がりの場合で90°を超えない場合、表示値は逆転しません。

a. 勾 配

逆転

（図１２）

勾配

1/200

1/150

1/115

1/100

角度

0.3°

0.4°

0.5°

0.6°

割ピッ

b. 角 度

11．
１５秒ほど後に「ピッ」と音が鳴り、［ＣＡＬ５］が表示されます。
（図７）
その後［ＵＰ］←→［-６-］が切り替り表示されます。
12．
本器をそのまま水平方向 に１８０°回転し、垂直面にあてます。
13．
「ピッ」と音が鳴り、［ＣＡＬ６］が表示されます。（図８）
その後［ＵＰ］←→［-７-］が切り替り表示されます。

（図１１）

勾配換算表

〔垂直の校正〕10．
大体垂直な面に、本器のデジタル表示部が上になるようにベース基準面をあてます。

14．
つづいて、本器のデジタル表示部が下になるようにベース基準面を垂直面にあてます。

0.00°

6（図４）

約 68°
逆転°

９．
「ピッ」と音が鳴り、［ＣＡＬ４］が表示されます。
（図６）
その後［ＵＰ］←→［-５-］が切り替り表示されます。

90° 22°ピッ68

８．本器をそのまま同一面上で水平方向に１８０°回転します。

〔角度の例〕

8°（図３）

６．つづいて、本器のケース上面を基準面にします。

７．
「ピッ」と音が鳴り、［ＣＡＬ３］が表示されます。
（図５）
その後［ＵＰ］←→［-４-］が切り替り表示されます。

表示の逆転について

6 8°

５．
「ピッ」と音が鳴り、［ＣＡＬ２］が表示されます。
（図４）
その後［ＵＰ］←→［-３-］が切り替り表示されます。

５．本器を分解することは絶対にしないでください。

ピッ

３．１５秒ほど後に「ピッ」と音が鳴り、［ＣＡＬ１］が表示されます。
（図３）
その後［ＵＰ］←→［-２-］が切り替り表示されます。

４．本器をそのまま同一面上で水平方向 に１８０°回転します。

⑩表示保持ボタン
１回押すとその時の表示値を固定します。再度ボタンを押すと表示値の固定は解除されます。

ご注意

正２．１５秒ほど待った後に ［校正／ゼロ］
ボタンを２秒間押します。表示が ［ＣＡＬ］に変わったあと、［ＵＰ］←→［-１-］が切り替り表示されることを確認します。

⑨ ⑩⑨照明ボタン
１回押すとバックライトが点灯し、再度押すと消灯します。
また、点灯した状態にしておくと約１分で自動消灯します。

任意ゼロ設定〔水平の校正〕１．大体水平な定盤等の平面上へ、異物等が無いことを確認してか
ら本器のベース基準面を置き、［ＯＮ／ＯＦＦ］
ボタンを押し電源を入れます。

⑧⑧校正／ゼロボタン
水平及び、垂直の校正時や、任意ゼロ設定をする時に使用します。（詳細は「任意ゼロ設定」、「校正」をご参照ください。）＊精度が確認できたら、そのまま測定を開始してください。

校⑦ブザーボタン⑤⑥ ⑦
１回押すと 0°、90°、0％、0mm/Mの時にブザー音が鳴るように設定されます。再度ボタンを押すと設定は解除されます。

合

角 度
割 合

0.5％

0.0°

0.0％

〔角度例〕 （0％）（17.6％）
90° 80°

（図７）
ピッ

CAL７

0.7％

0.9％

10.0°

17.6％

1.0％

20.0°

36.4％

45°
（100％）

1/80

1.3％

0.7°

30.0°

57.7％

1/50

1.1°

45.0°

100％

3.3％

1.4°

60.0°

57.7％

1.9°

70.0°

36.4％

1/20

5.0％

1/10

2.9°

80.0°

17.6％

10.0％

5.7°

90.0°

0.0％

（10％＝0.1）

30°（57.7％）
20°（36.4％）

CAL８

1/30

2.5％

〔割合例〕

（図８）
ピッ

1/40

2.0％

1/1011010°（17.6％）

16．
本器をそのまま水平方向に１８０°回転し、垂直面にあてます。

0°（0％）

17．
「ピッ」と音が鳴り、［ＣＡＬ８］が表示されます。
（図１０）＊以上で、校正が完了しました。

＊ 0°〜 45°
（100％）までは水平に対し垂直の割合を表示します。
また、45.1°〜 90°（0％）までは垂直に対し水平の割合を表示します。

（図９）

（図１０）

I426-K 2011