PREPARATION FOR USE

INSTRUCTION MANUAL
Model No. TVC- 60/ TVC-100

PRECISION MEASURING TOOLSVERNIER CALIPERS●Loosen the lock screw.

Thank you for purchasing the Vernier Calipers.
This product is a precision measurement tool with a main scale and a vernier scale to indicate the measured value.

●For safe and proper use of this product, please read this instruction manual before use and follow the procedures described. Pleasekeep manual where it is accessible to user for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact distributor or place of purchase.

■PART IDENTIFICATION AND FUNCTION※Model TVC-60 shown.

⑤ Measuring surfaces for internal measurements㻝㻞 㻝㻟 㻝㻠 㻝㻡 㻝㻢 㻝㻣 㻝㻤 㻝㻥 㻞㻜 㻞㻝 ⑥
㻞㻞Lock㻞㻟Screw㻞㻠 㻞㻡 㻞㻢 ⑦㻞㻣Main㻞㻤Scale㻞㻥 㻟㻜 㻟㻝 㻟㻞 㻟㻟 㻟㻠 㻟㻡 㻟㻢 㻟㻣 㻟㻤④ Inside Jaws

③ Measuring surfaces forstep measurements

㻜

㻝㻞

㻟

㻠

㻡

㻢

㻣

㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻤㻥㻝㻞 㻠㻞㻝㻟 㻠㻟㻝㻠 㻠㻠㻝㻢 㻠㻢㻝㻣 㻠㻣㻝㻤 㻠㻤㻝㻥 㻠㻥㻞㻝 㻡㻝㻞㻞 㻡㻞㻞㻟 㻡㻟㻞㻠 㻡㻠㻞㻢 㻡㻢㻞㻣 㻡㻣㻞㻤 㻡㻤㻞㻥 㻡㻥㻟㻝 㻟㻞㼏㼙㻟㻟 㻟㻠 㻟㻡 㻟㻢 㻟㻣 㻟㻤 㻟㻥 㻠㻜㻝㻜 㻝㻝 㻠㻝㻝㻡 㻠㻡㻞㻜 㻡㻜㻞㻡 㻡㻡㻟㻜 㻢㻜㻜㻚㻜㻡㼙㼙⑧ Thumb Knob
② Outside Jaws

Moving the vernier slide with the lock screw tightened may cause excessive force to the slide which will damage the caliper and affectaccuracy.

●Wipe off any corrosion, dirt, or oil from measuring surfaces and slide.
Contamination of surfaces may cause measurement error.

●Allow the caliper and object to be measured time to reach the same temperature.
A temperature difference between the object to be measured and the caliper may cause measurement error. Allow enough time forthe temperatures to equilibrate.

●Performing an accuracy check.

①Jaw Alignment
In closed position, make sure the measuring surfaces on the outside jaws
㻟㻥 align㻠㻜 properly and there is no gap (no light should leak past.)②Confirm “0” points on scale (see diagram at right)
With jaws closed as in step ① above, make sure the reading is zero andthe vernier scale graduations align at the “0” and “39” positions as shown.
③Depth Confirmation
Position caliper on a flat surface for depth measurement. Reading shouldbe “0” as in step ② above, with the verier graduations aligned at the “0”and “39” positions.

⑨Vernier Slide ( Vernier Scale)① Measuring surfaces for outside measurements①Measuring Surfaces for outside measurements … Object to be measured is positioned between these surfaces.
②Outside Jaws ………………………………………… Measures extremal dimensions. Move Slide to open and close.
③Measuring surfaces for step measurements …… Measures the height of steps.
④Inside Jaws…………………………………………… Measures internal dimensions. Move Slide to open and close.
⑤Measuring surfaces for internal measurements … Position surfaces inside object to be measured.
⑥Lock Screw …………………………………………… Locks the Vernier Slide position.
⑦Main Scale …………………………………………… Indicates measured value in 1mm graduations.
⑧Thumb Knob ………………………………………… Position thumb on knob for precise control of vernier slide.
⑨Vernier Slide (Vernier Scale) ……………………… Scale for measuring with a resolution of 0.05mm.
Refer to following page for usage instructions.

SAFETY PRECAUTIONS
Always follow the proper operating procedures in order to prevent
harm to yourself or others, and to prevent damage to property.

WARNING

Before using the caliper, follow the steps below:

Denotes a requirement You MUST do

Indicates risk of personal injury or property damage if not followed.

Read the manual and follow the directions.
・Use of product other than as described in the manual
may cause accident.
Use only for measuring.
・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.
Handle with Care.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・Do not scratch instrument, for example by writing ID
number.

Use In Proper Environment.
・Please do not use instrument in rain or locations withexcessive humidity. Do not use at extreme high or low
temperatures. Keep out of direct sunlight.

Do not disassemble or modify.
・Do not remove the vernier slide, Please do not attempt todisassemble or modify as it may cause damage or poor accuracy.

CALIBRATION●In order to maintain instrument accuracy, it is recommended thataccuracy is confirmed through calibration on a periodic basis.
Wear of measuring surfaces from repeated use may affect accuracy and periodic
accuracy checks should be performed.

We provide calibration services.
Please contact agent in countryof purchase to makearrangements.

㻟㻠㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻡㻢㻣㻜㻚㻜㻡㼙㼙

Proper alignment

The measured value is determined by reading the main scale,and then adding the Vernier Scale reading as determined by
the position where the vernier and main scale graduations align.

In order to prevent measurement error, please note the following.

Measured Value = Main Scale Reading + Vernier Scale Reading●EXAMPLE 1㻡㻢

Main Scale(1mm)Measured Value: 77.00mm

㻣Vernier Scale(0.05mm)

㻤㻢Main Scale(1mm)㻣㻢Vernier Scale(0.05mm)㻣㻜㻚㻜㻡㼙㼙

77＋ 0.0 0
7 7. 0 0

Measured Value: 77.30mm㻤㻥㻜㻚㻜㻡㼙㼙

77＋ 0.3 0
7 7. 3 0

Measured Value: 77.35mm

㻤㻥

POSITIONING CALIPER ON OBJECT＜Flat surfaces＞

Jaw

When measuring featureswith flat surfaces, make surethe measuring surface of the㻝㻣jaws are parallel to thesurfaces to be measured.

＜Cylindrical surfaces＞
When measuring cylindricalfeatures, make sure thecaliper jaws are perpendicular
to the cylinder axis as shown.

Object being measured㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠㻝㻡 㻝㻢 㻝㻣Reading㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜●EXAMPLE 3㻡㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠㻝㻡 㻝㻢Reading㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜Vernier Scale(0.05mm)

Main Scale(1mm)㻥㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠㻝㻡 㻝㻢Reading㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻜㻚㻜㻡㼙㼙

77＋ 0 .3 5
7 7 .3 5

Jaw Tips Are Sharp̶Handle With Care.
・Careless handling may cause injury.
Do not move the Vernier Slide when the Lock Screwis tightened.
・Excessive force applied to Slide may cause damage oraffect accuracy.

㻞PREVENTING ERRORS㻡Denotes a prohibition You MUST NOT do㻝

READING THE SCALE●EXAMPLE 2

Please Observe

㻜VIEWING THE SCALE

Scale should be read fromdirectly above the calipers.
Due to step in height of vernierscale relative to main scale, if
viewing direction is not directly
above the reading may have㻝㻣parallax error.

SPECIFICATIONSAFTER USE CARE, STORAGE NOTES●As required, wipe slide and measuring surfaces with adry cloth treated with anti-corrosive oil. When not in use,
apply anti-corrosive treatment or keep in anti-corrosivebag.
●When not in use, keep jaws slightly open and lock screw
slightly loose to prevent over tightening due to thermal
expansion which may affect accuracy.
●Store in supplied case in a cool, dry location.

●Material : Stainless Steel
Model No.
Measuring Range
Resolution
Maximum permissible error
WeightTVC-60
600mm

TVC-100
1000mm0.05 mm±0.10mm±0.15mmapprox. 1.2 kg approx. 2.5kg

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

使用前の準備

取扱説明書 品番 TVC- 60/TVC-100

精密測定工具シルバー標準型ノギス

ご使用前に、下記の準備を行ってください。

●止めねじをゆるめる。

止めねじを締めたままバーニヤを動かすと、無理な力がかかり、精度不良や破損の原因となります。

この度は「シルバー標準型ノギス」をお買上げいただきありがとうございます。
この商品は、本尺目盛と副尺（バーニヤ）目盛を使い、測定値を読み取る精密測定工具です。

●測定面、摺動面、外観などの汚れや防錆油などを拭き取る。

●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称と機能※イラストはTVC-60です。

⑤内側用測定面⑥止めねじ㻜㻝㻞㻟㻠㻡㻢㻣㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻤㻥●本体と被測定物の温度慣らしを十分に行う。

測定器本体と被測定物の間に温度差があると、誤差の原因となります。同じ温度になるよう、温度慣らしは十分に行ってください。

●使用前精度確認を行う。

①ジョウの確認
外側用ジョウを閉じた状態で外側用ジョウの部分を光に透かして、隙間がないか（光が漏れていないか）確認してください。

⑦本尺（本尺目盛）④内側用ジョウ
③段差測定面そのまま測定を行うと誤差の原因となります。

②副尺の０点が合致しているか（右図）①の状態で、副尺目盛の２つの０が、本尺目盛の０と３９に合致していることを確認してください。

㻜㻚㻜㻡㼙㼙

①外側用測定面 …… 被測定物をこの面で挟んで測定します。

②外側用ジョウ …… 被測定物の外側を測ります。バーニヤをスライドさせると開閉します。

③段差測定面 ……… 段差面に当てて長さを測定します。

④内側用ジョウ …… 被測定物の内側を測ります。バーニヤをスライドさせると開閉します。

⑤内側用測定面 …… 被測定物の内側にこの面を当てて測定します。

安全上のご注意⑥止めねじ …………… バーニヤのスライドを固定します。

本尺目盛に、本尺目盛と副尺（バーニヤ）目盛が合致したところの副尺目盛をプラスして、測定値を読みます。

測定値＝本尺目盛＋副尺（バーニヤ）目盛⑦本尺（本尺目盛）…… １ｍｍ単位を測定する目盛です。

⑧指かけ ……………… ここに親指をかけてバーニヤをスライドさせます。

⑨バーニヤ …………… ０.０５ｍｍ単位を測定する目盛です。
目盛の読み方は、右記を参照してくだ（副尺目盛）さい。

してはいけない内容『禁止事項』です。

必ず実行していただく内容『強制事項』です。

取扱説明書をよく読み、指示に従う。
・取扱説明書に記載された内容以外での使用は、事故の原因となります。

使用環境に注意する。
・雨の中や湿気の多い場所、高温・低温環境下、直射日光の当たる場所での使用はおやめください。

測定作業にのみ使用する。
・指定用途以外へのご使用は、製品の破損や摩耗、予測できない事故の原因となります。

ジョウ先端は尖っているので、扱いに注意する。
・けがの原因となります。

校正について●測定精度を保つため、一定周期で校正による精度確認をおすすめします。

性能の劣化は、使用の頻度・環境によって大きく異なります。
お客様の社内規格に準じた周期で、定期点検を行うことをおすすめ致します。

㻢本尺目盛（1㎜）
副尺目盛（0.05㎜）

㻡

注 意 お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

大切に取り扱う。
・ぶつける、落とすなどの衝撃を与えたり、重量物を上に載せたりしないでください。精度不良や破損の原因となります。
・番号の書き込みなど、商品に傷をつけないでください。

㻡㻣㻤㻥

止めねじを締めた状態で、バーニヤをスライドさせない。
・無理な力がかかり、精度不良や破損の原因となります。
分解・改造を行わない。
・バーニヤを外したり、分解・改造を行わないでください。精度不良や破損の原因となります。

弊社にて校正を受け付けております。全国にある弊社代理店、取扱店を経由してお申し込みください。

お問い合わせ：( 0 2 5 6 ) 3 3 - 5 5 0 2㻢＋㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻣㻤㻥㻜㻚㻜㻡㼙㼙

770 .0 0
7 7.0 0㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻡㻢㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻣㻤㻥㻢㻡㻣㻜㻚㻜㻡㼙㼙

合致していること

測定誤差を防ぐため、下記の事項に注意してください。

被測定物への当て方

ジョウ＜直方体を測定する場合＞
平行する面に対して垂直にジョウ部分を当てます。
＜円筒体を測定する場合＞
円筒体の軸方向に対して、ノギスが直角になるようにジョウ部分を当てます。

被測定物

目盛の見方

㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠 読み㻝㻡 㻝㻢 㻝㻣㻜㻚㻜㻡㼙㼙

77＋ 0 .3 07 7.3 0●測定例３ 測定値 77.35㎜
本尺目盛
（1㎜）
副尺目盛
（0.05㎜）㻠㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠 読み㻝㻡 㻝㻢 㻝㻣●測定例２ 測定値 77.30㎜

必ずお守りください。

お使いになる人や他の人への危害、財産への損害を未然に防止するた
め、必ずお守りいただくことを右記のように説明しています。

●測定例１ 測定値 77.00㎜
本尺目盛
（1㎜）
副尺目盛
（0.05㎜）

㻟

誤差の防止

目盛の読み方⑨バーニヤ（副尺目盛）①外側用測定面

㻞③デプスの確認
深さ測定面を平らな面に垂直に当て、②と同様に副尺目盛の２つの０が、本尺目盛の０と３９に合致していることを確認してください。

㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠㻠㻝 㻝㻡㻠㻞 㻝㻢㻠㻟 㻝㻣㻠㻠 㻝㻤㻠㻡 㻝㻥㻠㻢 㻞㻜㻠㻣 㻞㻝㻠㻤 㻞㻞㻠㻥 㻞㻟㻡㻜 㻞㻠㻡㻝㻞㻡㻡㻞㻞㻢㻡㻟㻞㻣㻡㻠㻞㻤㻡㻡㻞㻥㻡㻢㻟㻜㻡㻣 㻟㻝㻡㻤 㻟㻞㻡㻥 㻟㻟㻢㻜 㻟㻠 㼏㼙㻟㻡 㻟㻢 㻟㻣 㻟㻤 㻟㻥 㻠㻜⑧指かけ②外側用ジョウ㻝㻜

ノギスの目盛のある方向に垂直に読み取ります。垂直でないと本尺と副尺（バーニヤ）の面の高さの段差により、視差ができ誤差が発生します。

㻝㻜 㻝㻝 㻝㻞 㻝㻟 㻝㻠 読み㻝㻡 㻝㻢 㻝㻣㻜 㻝 㻞 㻟 㻠 㻡 㻢 㻣 㻤 㻥 㻜㻜㻚㻜㻡㼙㼙

77＋ 0 .3 5
7 7.3 5

使用後のお手入れ・保管方法●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う。

防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

●止めねじをゆるめ、ジョウを少し開いた状態で保管する。

製品仕様●材質：ステンレススチール
品

番

最大測定長
最小読取値

最大許容誤差質量

TVC-60

TVC-100

600 mm±0.10mm
約 1.2 kg

1000mm0.05 mm±0.15mm
約 2.5kg

ジョウを閉じた状態で止めねじを締めて固定すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

●付属のケースに入れ、乾燥した冷暗所に保管する。

直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp2312
I466-K㻤