Multiple Application Platform
FVAi-2030 Benchtop Microscope
User Guide
22112369-339 R002, Standard
November 2023

Copyright
Copyright © 2023 VIAVI Solutions Inc. All rights reserved. VIAVI and FiberChekPRO™ are trademarks or registeredtrademarks of VIAVI in the United States and/or other countries. All other trademarks and registered trademarks
are the property of their respective owners. No part of this guide may be reproduced or transmitted,
electronically or otherwise, without written permission of the publisher.
Every effort was made to ensure that the information in this manual was accurate at the time of printing.
However, information is subject to change without notice, and VIAVI reserves the right to provide an addendum
to this manual with information not available at the time that this manual was created.
Patents
Patented as described at www.viavisolutions.com/patents.
Terms and Conditions
Specifications, terms, and conditions are subject to change without notice. The provision of hardware, services,
and/or software are subject to VIAVI standard terms and conditions, available at www.viavisolutions.com/terms.
Disclaimer
The foregoing warranties are in lieu of all other warranties and conditions, expressed or implied. VIAVI specificallydisclaims all other warranties, either expressed or implied, on any hardware, documentation or services including
but not limited to warranties relating to quality, performance, non-infringement, merchantability or fitness for a
purpose, as well as those arising from any course of dealing, usage or trade practice. Under no circumstances will
VIAVI be liable for any indirect or consequential damages related to breach of this warranty.
California Proposition 65
California Proposition 65, officially known as the Safe Drinking Water and Toxic Enforcement Act of 1986, wasenacted in November 1986 with the aim of protecting individuals in the state of California and the state's drinking
water and environment from excessive exposure to chemicals known to the state to cause cancer, birth defects orother reproductive harm. For the VIAVI position statement on the use of Proposition 65 chemicals in VIAVI
products, see the Hazardous Substance Control section of the VIAVI Policies & Standards web page.
Federal Communications Commission (FCC) NoticeElectronic test equipment is exempt from Part 15 compliance (FCC) in the United States.
Product Environmental Compliance
VIAVI is committed to compliance with all applicable laws and regulations controlling the use of hazardous
substances in its products, as well as the disposal of equipment (including batteries) and waste packaging. For
details, see the VIAVI Policies & Standards web page or contact the VIAVI WEEE Program Management team at
Global.WEEE@ViaviSolutions.com.
EU REACH
Article 33 of EU REACH regulation (EC) No 1907/2006 requires article suppliers to provide information if a listed
Substances of Very High Concern (SVHC) is present in an article above a certain threshold. For information on thepresence of REACH SVHCs in VIAVI products, see the Hazardous Substance Control section of the VIAVI Policies &Standards web page.
EU CE Marking Directives (LV, EMC, RoHS, RE)
This product conforms with all applicable CE marking directives. Please request the EU Declaration of Conformity
for details.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standardii

China RoHS materials declaration
1445 South Spectrum Blvd., Suite 102
Chandler, AZ 85286, USA

22157638, Rev 001“ѝѝ ഭ RoHS”
ǉ⭥ᆀؑ᚟ӗ૱⊑ḃ᧗ࡦ㇑⨶࣎⌅Ǌ˄ؑ᚟ӗъ䜘ˈㅜ 39 ਧ˅䱴ᖅ (Additional Information required for the Chinese Market only)ᵜ䱴ᖅ᤹➗Āѝഭ RoHSāⲴ㾱≲䈤᰾Ҷᴹ‫⭥ޣ‬ᆀؑ᚟ӗ૱⧟‫⭘֯؍‬ᵏ䲀ⲴᛵߥˈᒦࡇࠪҶӗ૱ѝਜ਼ᴹ
Ⲵᴹ∂ǃᴹᇣ⢙䍘Ⲵ⿽㊫઼ᡰ൘䜘ԦǄᵜ䱴ᖅ䘲⭘Ҿӗ૱ѫփ઼ᡰᴹ䝽ԦǄӗ૱㌫ࡇ: FVAi Series Bench-Top Microscope with Integrated LCD Autofocus 30x(Product Family)⧟‫⭘֯؍‬ᵏ䲀˖ᵜḷ䇶ḷ⌘Ҿӗ૱ѫփѻкˈ㺘᰾䈕ӗ૱ᡆަ䝽Ԧਜ਼ᴹᴹ∂ǃᴹᇣ⢙䍘˄䈖ᛵ㿱л㺘˅ǄަѝⲴᮠᆇԓ㺘൘↓ᑨ᫽֌ᶑԦл㠣ቁ൘ӗ૱⭏ӗᰕᵏѻਾᮠᒤ޵䈕ӗ૱ᡆަ䝽Ԧ޵
ਜ਼ᴹⲴᴹ∂ǃᴹᇣ⢙䍘нՊਈᔲᡆ⋴┿Ǆ䈕ᵏ䲀н䘲⭘Ҿ䈨ྲ⭥⊐ㅹ᱃㙇૱Ǆ
ᴹ‫↓ޣ‬ᑨ᫽֌ᶑԦˈ䈧৲㿱ӗ૱⭘ᡧ᡻޼Ǆӗ૱⭏ӗᰕᵏ䈧৲㿱ӗ૱Ⲵ৏࿻ṑ߶䇱ҖǄ

ᴹ∂ǃᴹᇣ⢙䍘Ⲵ㊫ර઼ᡰ൘䜘Ԧ
ᴹ∂ǃᴹᇣ⢙䍘઼‫ݳ‬㍐
‫ݳ‬ಘԦ

䫵(Pb)

⊎ (Hg)

䭹(Cd)

‫ޝ‬ԧ䬜
(CR6+)

ཊⓤ㚄㤟
(PBB)

ཊⓤҼ㤟䟊
(PBDE)

X

O

O

O

O

O

X

O

O

O

O

O

(Component)ӗ૱ѫփ(Main Product)ঠࡧ⭥䐟ᶯ㓴Ԧ(PCB Assemblies)޵䜘䝽㓯(Internal wiring)ᱮ⽪ಘ(Display)䭞ⴈ(Keyboard)⭥⊐(Batteries)⭥Ⓚ(Power Supply)c ⭥ᐕ䴦Ԧ(Electro-mechanical parts)

⺜ⴈ(Hard Drive)‫⁑ݹ‬ඇ / 䖵ࣙ⁑ඇ(Optical modules) / (Auxiliary modules)䠁኎ཆ༣䴦Ԧ઼㍗ᢓԦ
(Metal case parts and fixings)

ກᯉཆ༣䴦Ԧ
(Plastic case parts)ḷㆮ઼㜦ᑖ(Labels and tapes)

X

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

X

O

O

O

O

O

X

O

O

O

O

O

O

O

O

O

O

O

X

O

O

O

O

O

X

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

䝽Ԧ
(Accessories)ཆ᧕⭥㔶઼䘲䝽ಘ(External cables and adapters)᡻޼઼ަᆳঠࡧᶀᯉ
(Handbooks and other printed material)व㻵㇡઼㕊ᑖ
(Carrying case and strap)ަᆳ䝽Ԧ(Other accessories)

X

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

O

ᵜ㺘ᱟ᤹➗ S / T 11364 Ⲵ㿴ᇊ㕆ࡦⲴ:O˖㺘⽪䈕ᴹᇣ⢙䍘൘䈕䜘Ԧᡰᴹ൷䍘ᶀᯉѝⲴਜ਼䟿൷൘ GB/T 26572 㿴ᇊⲴ䲀䟿㾱≲ԕлǄ
X˖㺘⽪䈕ᴹᇣ⢙䍘㠣ቁ൘䈕䜘ԦⲴḀа൷䍘ᶀᯉѝⲴਜ਼䟿䎵ࠪ GB/T 26572 㿴ᇊⲴ䲀䟿㾱≲Ǆ

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standardiii

Contents
1

FVAi-2030 Benchtop Microscope .................................................................................................1-1
Ordering information.......................................................................................................................... 1-2
Technical Assistance Center and Knowledge Base ............................................................................. 1-3

2

Getting started ............................................................................................................................2-1
Features .............................................................................................................................................. 2-2
Powering the FVAi-2030 microscope on or off................................................................................... 2-3
Mounting fiber adapters and connectors........................................................................................... 2-4
Navigating the Home screen .............................................................................................................. 2-6
Accessing Live view............................................................................................................................. 2-8
Applying a profile and an optical setting .......................................................................................... 2-10
Managing groups .............................................................................................................................. 2-12
Managing user information .............................................................................................................. 2-14
Managing microscope settings ......................................................................................................... 2-15
Managing the Fiber Centering Limit setting and marker ........................................................... 2-16
Managing image-save options.......................................................................................................... 2-18
Managing system settings ................................................................................................................ 2-19
Calibrating the touchscreen ....................................................................................................... 2-20
Connecting and activating a supported USB device ......................................................................... 2-21
Activating a P5000i probe .......................................................................................................... 2-21
Activating an MP Series optical power meter............................................................................ 2-23

3

Fiber inspection during standalone operation..............................................................................3-1
Requirements for Pass/Fail tests ........................................................................................................ 3-2
Starting a Pass/Fail test from the FVAi-2030 ...................................................................................... 3-3
Viewing the fiber chamfer .................................................................................................................. 3-4
Capturing images ................................................................................................................................ 3-5
Recalling test results and images........................................................................................................ 3-7

4

Using the FVAi-2030 microscope with FiberChekPRO...................................................................4-1
Connecting to FiberChekPRO.............................................................................................................. 4-2
Performing Pass/Fail tests and generating reports ............................................................................ 4-3
Viewing the fiber chamfer with FiberChekPRO .................................................................................. 4-5
Creating analysis profiles for the FVAi-2030 microscope ................................................................... 4-6

Appendix A Firmware update ........................................................................................................... A-1
Appendix B System reset .................................................................................................................. B-1
Appendix C Specifications ..................................................................................................................C-1
Appendix D Document revision history ............................................................................................ D-1

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standardiv

About this guideThis guide provides information about the FVAi-2030 Benchtop Microscope. The following updates appear in thisrevision:•

Added the section “Safety instructions”•

Updated the section “Accessing Live view”

For a detailed revision history, see “Document revision history” on page D-1.
To access the latest version of this user guide, go to:
FVAi-2030 Benchtop Microscope User Guide, 22112369-339

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standardv

Safety instructions
CAUTION
Turn off the equipment, and disconnect all cables connected to it before moving the equipment or
performing maintenance procedures.

ATTENTION
Éteignez l'équipement et débranchez tous les câbles qui y sont connectés avant de déplacerl'équipement ou d'effectuer des procédures de maintenance.

WARNING• Only use power cords with adequate ratings and grounding-type plugs. Contact VIAVI for replacement
power cords.
• Do not alter a power cord by modifying the grounding-type plug or using an adapter.
• Do not allow anything to rest on the power cord, and do not locate the equipment where persons canwalk on the power cord.
• The equipment is rated for indoor use only. Do not use the equipment outdoors.
• To prevent potential fire or shock hazard, do not expose the equipment to any source of excessive
moisture.
• Do not perform any operating or maintenance procedure that is not described in the user
documentation. If the equipment is used in a manner not specified by VIAVI, the protection provided
by the equipment might be impaired.
• Do not attempt to service this product yourself, as opening or removing covers might expose you todangerous high-voltage points and other hazards. Refer all servicing to qualified VIAVI service
personnel.
• Do not operate any equipment with its covers or panels removed.

AVERTISSEMENT• Utilisez uniquement des cordons d'alimentation avec des caractéristiques nominales adéquates etdes fiches de type mise à la terre. Contactez VIAVI pour les cordons d'alimentation de remplacement.
• Ne modifiez pas un cordon d'alimentation en modifiant la prise de type mise à la terre ou en utilisant
un adaptateur.
• Ne laissez rien reposer sur le cordon d'alimentation et ne placez pas l'équipement où des personnes
peuvent marcher sur le cordon d'alimentation.
• L'équipement est conçu pour une utilisation en intérieur uniquement. Ne pas utiliser l'équipement àl'extérieur.
• Pour éviter tout risque d'incendie ou d'électrocution, n'exposez pas l'équipement à une sourced'humidité excessive.
• N'effectuez aucune procédure d'exploitation ou de maintenance qui n'est pas décrite dans la
documentation utilisateur. Si l'équipement est utilisé d'une manière non spécifiée par VIAVI, laprotection fournie par l'équipement peut être altérée.
• N'essayez pas de réparer ce produit vous-même, car l'ouverture ou le retrait des couvercles peut vous
exposer à des points haute tension dangereux et à d'autres dangers. Confiez tous les travauxd'entretien au personnel de service qualifié de VIAVI.
• N'utilisez aucun équipement dont les couvercles ou les panneaux ont été retirés.

FVAi-2030 Benchtop Microscope User Guide

22112369-339 R002, Standard, vi

1 FVAi-2030 Benchtop Microscope
The FVAi-2030 Benchtop Microscope is the ideal microscope to control contamination during optical connector
manufacturing. Equipped with 30x optics, ultra-wide field-of-view, dual magnification, automatic focus, and builtin end face analysis, the FVAi-2030 gives operators fast and complete visibility of the entire ferrule end face sothat any contamination on or near the fiber end face can be quickly found.
Operators can use the FVAi-2030 as a standalone device or in conjunction with the FiberChekPRO™ inspection
and analysis software.
Figure 1-1: FVAi-2030 Benchtop Microscope

FiberChekPRO Fiber Inspection and Analysis software
Designed to operate on a PC/laptop, FiberChekPRO instantly captures, analyzes, and grades fiber end-face images
to determine acceptability of the end faces. It identifies and characterizes defects and contamination, and thenprovides a pass or fail result according to preconfigured criteria. In addition, users can customize settings, as well
as archive data and generate reports.
This section covers the following information:•“Ordering information” on page 1-2•“Technical Assistance Center and Knowledge Base” on page 1-3

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

1-1

FVAi-2030 Benchtop Microscope

Ordering information

Ordering information
Table 1-1: FVAi-2030 Benchtop Microscope
Description

Part Number

FVAi-2030 Large Field of View Benchtop Microscope1 with Integrated LCD, Autofocus 30x

FVAi-2030

1. Includes M5 hex screwdriver, power adapter, and USB cable.

Table 1-2: Supported fiber adapters
Description

Part Number

LC/APC Duplex Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-LCA-DUX-SL35

LC/APC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-LCA-SL35

LC/PC Duplex Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-LC-DUX-SL35

LC/PC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-LC-SL35

SC/APC Fiber Microscope Adapter, 30-Degree Chamfer Angle

FMAX-SCA-SL30

SC/APC Fiber Microscope Adapter, 41-Degree Chamfer Angle

FMAX-SCA-SL41

SC/PC Duplex Fiber Microscope Adapter. 30-Degree Chamfer Angle

FMAX-SC-DUX-SL30

SC/PC Fiber Microscope Adapter, 30-Degree Chamfer Angle

FMAX-SC-SL30

Universal 1.25mm APC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-U12A-SL35

Universal 1.25mm PC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-U12-SL35

Universal 2.5mm APC Fiber Microscope Adapter, 30-Degree Chamfer Angle

FMAX-U25A-SL30

Universal 2.5mm APC Fiber Microscope Adapter, 41-Degree Chamfer Angle

FMAX-U25A-SL41

Universal 2.5mm PC Fiber Microscope Adapter, 30-Degree Chamfer Angle

FMAX-U25-SL30

CS/APC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-CS-APC-SL35

CS/PC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-CS-SL35

MDC/APC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-MDC-APC-SL35

MDC/PC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-MDC-SL35

SN/APC Duplex Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-SN-APC-SL35

SN/PC Fiber Microscope Adapter, 35-Degree Chamfer Angle

FMAX-SN-SL35

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

1-2

FVAi-2030 Benchtop MicroscopeTechnical Assistance Center and Knowledge Base

Technical Assistance Center and Knowledge BaseTo find the Technical Assistance Center phone number and email in your region for the FVAi-2030 Benchtop
Microscope or to search the VIAVI Solutions Knowledge Base, visit the VIAVI Solutions Technical & ProductSupport site at support.viavisolutions.com.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

1-3

2 Getting started
This section covers the following information:•“Features” on page 2-2•“Powering the FVAi-2030 microscope on or off” on page 2-3•“Mounting fiber adapters and connectors” on page 2-4•“Navigating the Home screen” on page 2-6•“Accessing Live view” on page 2-8•“Applying a profile and an optical setting” on page 2-10•

“Managing groups” on page 2-12•

“Managing user information” on page 2-14•

“Managing microscope settings” on page 2-15•

“Managing image-save options” on page 2-18•

“Managing system settings” on page 2-19•

“Connecting and activating a supported USB device” on page 2-21

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-1

Getting started

FeaturesFeatures
Figure 2-1: FVAi-2030 microscope features (without adapter)

Table 2-1: FVAi-2030 microscope features - rear panel
Label

Description
Power switch (ON/OFF). See “Powering the FVAi-2030 microscope on or off” on page 2-3).
Power adapter receptacle
USB-B Interface (1x) for connecting the FVAi-2030 microscope to a computer for use with FiberChekPRO
inspection software. See “Using the FVAi-2030 microscope with FiberChekPRO” on page 4-1.
USB Interfaces (2x) for connecting supported USB devices to FVAi-2030 microscope. See “Connecting andactivating a supported USB device” on page 2-21.
VGA Interface for connecting the FVAi-2030 microscope to an external monitor.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-2

Getting started

Powering the FVAi-2030 microscope on or off

Powering the FVAi-2030 microscope on or off
CAUTIONUse only the power adapter shipped with the FVAi-2030 microscope to power thedevice.
Note: This section describes the procedure for powering on the microscope for standalone use. For information
about powering the microscope for use with the FiberChekPRO inspection software, see “Using the FVAi2030 microscope with FiberChekPRO” on page 4-1.
Step 1

Place the microscope on a flat, stable surface in a location close to an appropriate power source.

Step 2Connect the power adapter to the receptacle on the rear panel of the microscope and then to the powersource.

Step 3

Set the power switch on the rear panel to the ON position.
The LCD touchscreen initializes.
If the microscope is not connected to a computer via a USB-B connection (see “Features” on page 2-2),
the Home screen appears.

If the microscope is connected to a computer via a USB-B connection, select the Home tile to access theHome screen. For more information, see “Navigating the Home screen” on page 2-6.

Step 4

To power off the microscope, set the power switch to the OFF position.
Important: Always use the power button to power the microscope off. Do not power off bydisconnecting the power cable from either the device or the power source.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-3

Getting started

Mounting fiber adapters and connectors

Mounting fiber adapters and connectors
Important: Store fiber adapters in dust-proof containers while not installed on a microscope.
Step 1

If present, unscrew the lens cap or fiber adapter from the adapter mount, and store it in a dust-proof
container.

Step 2

Place the required fiber adapter onto the mount, and finger-tighten the adapter to secure it. Do not
over-tighten the adapter.

Step 3

Power on the microscope, and access Live view on the LCD touchscreen (see “Accessing Live view” onpage 2-8).

Step 4

Clean the fiber connector to be inspected.

Step 5

Ensure that the fiber connector is correctly oriented, and then insert it into the fiber adapter.

Step 6

Ensure that the fiber end face is properly centered within the view.

Note: The message Auto Focus Error: Unable to locate the fiber will appear when you
attempt to use the Autofocus tool while the fiber end face is poorly centered.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-4

Getting started

Step 7

Mounting fiber adapters and connectors

If required, use the M5 hex screwdriver supplied with the microscope to adjust the vertical andhorizontal position of the lens.
Note: You can use the Fiber Centering Limit setting and marker as a reference for the adjustment. For
information, see “Managing the Fiber Centering Limit setting and marker” on page 2-16.
Insert the end of the M5 hex screwdriver into an adjustment pilot hole until it engages, and thencarefully turn the screwdriver in either direction, referring to the touchscreen as the position of the lenschanges.

CAUTION• Use only the M5 hex screwdriver supplied with the microscope.
• Do not force the screwdriver beyond the adjustment limits.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-5

Getting started

Navigating the Home screen

Navigating the Home screen
Use the Home screen during standalone operation of the microscope to access tools on the FVAi-2030 microscope(Table 2-2 on page 2-7).
Note: If touchscreen controls (such as tiles, menu options, and keyboard keys) do not respond as expected, you
can calibrate the active areas of the touchscreen. See “Calibrating the touchscreen” on page 2-20 for
information.
Figure 2-2: FVAi-2030 Home screenNote:

If supported devices, such as a P5000i Digital Analysis Probe and/or an MP Series Optical Power Meter,are connected to the microscope via a USB connection (see “Features” on page 2-2), a tile for activating
the supported device appears on the Home screen. Select the tile for the device to use the device withthe microscope. See “Connecting and activating a supported USB device” on page 2-21 for more
information.

Figure 2-3: FVAi-2030 Home Screen example - connected USB devices

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-6

Getting started

Navigating the Home screen

Table 2-2: FVAi-2030 Home screen tools
Tool

Description
Inspect (FVAi-2030)

Activates the FVAi-2030 microscope and accesses Live view for fiber
inspection. See “Accessing Live view” on page 2-8.

More (Home Screen menu,
Figure 2-4)

RECALL IMAGE: See “Recalling test results and images” on page 3-7
REVIEW OPM DATA: See “Connecting and activating a supported USB
device” on page 2-21.
SYSTEM SETTINGS: See “Managing system settings” on page 2-19
USER INFO: See “Managing user information” on page 2-14
OUTPUT:• During standalone operation, OUTPUT:LOCAL indicates local
operation via the touchscreen.
• While the microscope is connected to an external monitor via a
VGA connection (OUTPUT:VGA), and/or to FiberChekPRO via a
USB-B connection to the computer where FiberChekPRO is
installed, select to toggle output mode for working with themicroscope.
Note: While output mode is FiberChekPRO over a USB-B connection,
select the Home tile on the LCD touchscreen to access the Home
screen on the touchscreen. See “Powering the FVAi-2030 microscopeon or off” on page 2-3.

Figure 2-4: Home Screen menu example

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-7

Getting started

Accessing Live view

Accessing Live view
During standalone operation of the microscope, the Live view on the touchscreen displays a live view of the fiberconnector mounted on the fiber adapter and provides access to various fiber inspection and management tools.
Note: For information about Live view features available in FiberChekPRO, see the FiberChekPRO AutomatedFiber Inspection and Analysis Software User Manual.
Step 1On the Home screen, select Inspect to access Live view.
The following image shows that a fiber connector is mounted on the microscope and the Focus-quality
meter is enabled.

Step 2

Tap the touchscreen to access the Live view tools, and then select a tool (Table 2-3 on page 2-9).

Note:

While the microscope is connected to an external monitor and in Live View, the touchscreen displays
only the Live view tools.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-8

Getting started

Accessing Live view

Table 2-3: Live view tools
Tool

Description

Home

Return to the Home screen (see “Navigating the Home screen” on page 2-6).

Magnification High/Low

Increase or decrease magnification of the image displayed on the touchscreen.

Start Test

See “Starting a Pass/Fail test from the FVAi-2030” on page 3-3.
Note: For information about starting a Pass/Fail test with FiberChekPRO, see“Using the FVAi-2030 microscope with FiberChekPRO” on page 4-1.

Chamfer Focus

See “Viewing the fiber chamfer” on page 3-4.
Note: For information about using FiberChekPRO to view the fiber chamfer, see“Using the FVAi-2030 microscope with FiberChekPRO” on page 4-1.

Autofocus

Automatically focus the view of the fiber end face on the touchscreen.
Note: If configured in the microscope settings, selecting Autofocus will trigger a
Pass/Fail test. See “Managing microscope settings” on page 2-15.

Inspection Settings menu

PROFILE: See “Applying a profile and an optical setting” on page 2-10.
OPTICAL SETTING: See “Applying a profile and an optical setting” on page 2-10.
CREATE NEW GROUP/CLOSE GROUP : GROUP NAME: See “Managing groups” onpage 2-12.
RECALL IMAGE1: See“Recalling test results and images” on page 3-7.
USER INFO1: See “Managing user information” on page 2-14.
MICROSCOPE SETTINGS: See “Managing microscope settings” on page 2-15.
IMAGE SAVE OPTIONS: See “Managing image-save options” on page 2-18.
SYSTEM SETTINGS1: See “Managing system settings” on page 2-19.
OUTPUT LOCAL1: See “Navigating the Home screen” on page 2-6.

CaptureSee “Capturing images” on page 3-5.

1. Also available on the Home Screen menu (see “Navigating the Home screen” on page 2-6).
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-9

Getting started

Applying a profile and an optical setting

Applying a profile and an optical setting
The FVAi-2030 microscope provides factory-set analysis profiles and optical settings for use in Pass/Fail testing(Table 2-4). A profile defines Pass/Fail thresholds for a fiber type, while an optical setting defines the lighting andsizing calibration applied during a test.
Refer to Table 2-4 to ensure that the appropriate profile and optical setting for the fiber connector are applied
before performing a test during standalone operation of the microscope.
Table 2-4: Factory-set analysis profiles and optical settings
With Analysis Profile...

Use Optical Setting...

FVAi-2030 SM 1.25 mm UPC (Polish detect)

FMAX-LC-SL35

FVAi-2030 SM 1.25 mm APC (Polish detect)

FMAX-LCA-SL35

FVAi-2030 SM 2.5 mm UPC (Polish detect)

FMAX-SC-SL30

FVAi-2030 SM 2.5 mm APC (Polish detect)

FMAX-SCA-SL30

FVAi-2030 MM50 1.25 mm UPC (Polish detect)

FMAX-LC-SL35

FVAi-2030 MM50 2.5 mm UPC (Polish detect)

FMAX-SC-SL30

Note: FiberChekPRO provides tools that let you create customized profiles (see “Creating analysis profiles for
the FVAi-2030 microscope” on page 4-6) and optical settings. You can also manage factory-set and custom
profiles and optical settings with FiberChekPRO, making them available on or removing them from themicroscope as required. For information, including information about applying profiles and opticalsettings using FiberChekPRO, see the FiberChekPRO Automated Fiber Inspection and Analysis Software
User Manual.
Step 1In Live view, select Inspection Settings, and then do either of the following:▪ Select PROFILE to access the profiles available on the microscope, select a profile, and then selectBack to return to the Inspection Settings menu.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-10

Getting started

Applying a profile and an optical setting▪ Select OPTICAL SETTINGS to access the optical settings available on the microscope, select an opticalsetting, and then select Back to return to the Inspection Settings menu.

Step 2

Select another option on the Inspection Settings menu, or select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-11

Getting started

Managing groups

Managing groups
Use groups during standalone operation of the microscope to organize and manage Pass/Fail data and captured
fiber images saved to the microscope. While a group is active, saved data and images are automatically added tothat group.
You can create a group and set it as the active group, or set an existing group as the active group. Closing theactive group allows you to save data and images without adding them to a group.
Note: For information about using and managing groups with FiberChekPRO, see the FiberChekPRO AutomatedFiber Inspection and Analysis Software User Manual.
Step 1In Live view, select Inspection Settings, and then do one of the following:▪ If the CLOSE GROUP : GROUP NAME option is available, select it to close the active group. The option
label changes to CREATE NEW GROUP.
Select Back to return to Live view, or select CREATE NEW GROUP to set an active group.

Note: The CREATE NEW GROUP option is available only while no other group is active.
▪ If the CREATE NEW GROUP option is available, select it to access the on-screen keyboard, and then go
to Step 2.

Step 2To set the active group, type either a unique name for the group you want to create or the name of an
existing group.
Note: The microscope provides a default group name (GROUP-001) that automatically increments. For
example, if you create a group using the default group name GROUP-001 and then save data or
images to it while it is active, the group name GROUP-002 is automatically provided as a option
when you close GROUP-001 and select CREATE NEW GROUP. For information about changing thedefault group base name, see “Managing image-save options” on page 2-18.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-12

Getting started

Step 3

Managing groups

Select OK to return to the Inspection Settings menu.
The option label changes to CLOSE GROUP : GROUP NAME, indicating that the specified group is active.

Step 4

Select another option, or select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-13

Getting started

Managing user information

Managing user information
During standalone operation of the microscope, you can add user information (for example, customer name,location, job ID, fiber ID, etc.) to help manage Pass/Fail data and fiber-analysis images saved to the microscope.
Fiber ID Base parameter
By default, a fiber ID is assigned to saved data and images and appears with each recalled image (see “Recalling
test results and images” on page 3-7). The fiber ID includes a prefix (base) and a value that increments as data andimages are saved; for example FIBER-001, FIBER-002, and so on. The default fiber ID base is FIBER; however, you
can change it as required when adding user information.
Note: For information about managing user information with FiberChekPRO, see the FiberChekPRO AutomatedFiber Inspection and Analysis Software User Manual.
Step 1In Live view, select Inspection Settings, and then select USER INFO.

Step 2On the User Information menu, scroll to and select an option to access an on-screen keyboard, type theinformation, and the click OK to return to the User Information menu.

The specified information appears beside the option label.
Step 3

Repeat Step 2 and Step 3 as required.

Step 4

Select Back to return to the Inspection Settings menu.

Step 5

Select another option, or select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-14

Getting started

Managing microscope settings

Managing microscope settings
Microscope settings let you specify operation parameters for the FVAi-2030 microscope.
Note:

For information about managing microscope settings (i.e., device settings) with FiberChekPRO, see theFiberChekPRO Automated Fiber Inspection and Analysis Software User Manual.

Step 1In Live view, select Inspection Settings, and then select MICROSCOPE SETTINGS.

Step 2

Specify microscope settings as required.

Setting

Description

AUTOCENTER

Enable (default) or disable automatic centering of the fiber image in Live view.

SHOW FOCUS METERShow (default) or hide the Focus-quality meter in Live view (see “Accessing
Live view” on page 2-8).

DETECT DEFECTS OUTSIDE ZONE

Enable or disable (default) detection of defects outside the largest inspection
zone.

AUTOFOCUS ON INSERT

Enable or disable (default) auto-focusing of the fiber image in Live view as thefiber connector is mounted on the fiber adapter.

TEST ON AUTOFOCUS

Enable or disable (default) automatic Pass/Fail test when Autofocus is selected(see “Accessing Live view” on page 2-8).

FIBER CENTERING LIMIT

Select to specify the fiber-centering-limit value using the on-screen keyboard.
(Range = 1 to 999 µm; default = N/A.) See “Managing the Fiber Centering Limit
setting and marker” on page 2-16).

SHOW CENTERING LIMITShow or hide (default) the Show Centering Limit marker in Live view (see“Managing the Fiber Centering Limit setting and marker” on page 2-16).

CHAMFER FOCUS OFFSET

Select to specify the chamfer focus offset value using the on-screen keyboard.
(Range = 0 to 2000 µm; default = 275 µm). The offset value specified adjusts
the focus on the chamfer view (see “Viewing the fiber chamfer” on page 3-4).

DISPLAY BUTTON MODE

Select to toggle the display mode to TEST (default) or CAPTURE.

AUTOFOCUS CALIBRATION

Recalibrate autofocus
Important: For use only at the direction and with the guidance of VIAVI
support.

PASS/FAIL UNLOCK

Not Applicable

Step 3

Select Back to return to the Inspection Settings menu.

Step 4

Select another option, or select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-15

Getting started

Managing microscope settings

Managing the Fiber Centering Limit setting and marker
The Fiber Centering Limit setting lets you define a centering limit for the fiber end face, which you can either
enable or disable by showing or hiding, respectively, a marker in Live view (Figure 2-5). The marker indicates the
position of the fiber end face relative to the defined centering limit. This is useful when centration limits are
required for Pass/Fail tests, and provides a helpful reference when manually adjusting fiber centering (see“Mounting fiber adapters and connectors” on page 2-4).
Figure 2-5: Fiber Centering Limit marker displayed in Live view

Important: While the specified fiber centering limit value is within the range of 1 to 999 µm and the option toshow the centering-limit marker is enabled, the message Inspection Error: Fiber is too far
out of center. Please re-center the fiber before testing will appear on thetouchscreen (Figure 2-6) when you attempt to start a Pass/Fail test on a fiber end face that is notaligned within the fiber-centering marker. Consequently, the test will not start.
Figure 2-6: Inspection Error message

Step 1In Live view, select Inspection Settings, select MICROSCOPE SETTINGS, and then select FIBER
CENTERING LIMIT to access the on-screen keyboard.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-16

Getting started

Step 2

Managing microscope settings

Enter a value within the specified range (1 to 999), and then select OK to return to the Microscope
Settings menu.

Note: Clearing the current value and returning to the Microscope Settings menu without specifying
another value automatically sets the value to N/A.
Step 3

Optionally, enable or disable SHOW CENTERING LIMIT.
Notes:• While SHOW CENTERING LIMIT is disabled, the specified fiber-centering limit (from 1 to 999) is notapplied.
• While the fiber-centering limit value is N/A, the marker will not be shown even while SHOWCENTERING LIMIT is enabled.

Step 4

Select Back to return to the Inspection Settings menu, and then select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-17

Getting started

Managing image-save options

Managing image-save options
Step 1In Live view, select Inspection Settings, and then select IMAGE SAVE OPTIONS.

Step 2

Specify settings as required.

Settings

Description

FIBER ID BASE

Select to edit the default fiber-ID prefix (base) using the on-screen keyboard.
For more information, see “Managing user information” on page 2-14.

GROUP NAME BASE

Select to edit the default group-name prefix (base) using the on-screen
keyboard. For more information, see “Managing groups” on page 2-12.

VERIFY NAME BEFORE SAVE

Select to confirm or change the fiber ID prefix before manually saving Pass/
Fail test results or a captured fiber-analysis image.

ONE BUTTON SAVE

Select to enable manual save of Pass/Fail test results and captured fiberanalysis images without requiring name verification.

SAVE RESULTS IF PASS

Select to enable Pass results to be automatically saved after a Pass/Fail test.

Note: The save options VERIFY NAME BEFORE SAVE, ONE BUTTON SAVE, and SAVE RESULTS IF PASS
are mutually exclusive. For example, selecting ONE BUTTON SAVE and then selecting SAVERESULTS IF PASS automatically disables ONE BUTTON SAVE.
Step 3

Select Back to return to the Inspection Settings menu.

Step 4

Select another option, or select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-18

Getting started

Managing system settings

Managing system settings
System settings include parameters such as language, date, time, and date and time formats, which are applied to
information saved with Pass/Fail test results and saved images. Also included are tools for calibrating thetouchscreen and retrieving system/device information
Step 1

Do one of the following:▪ On the Home screen, select More, and then select System Settings on the Home Screen menu (see“Navigating the Home screen” on page 2-6).
▪ In Live view, select Inspection Settings, and then select System Settings on the Inspection Settings
menu (see “Accessing Live view” on page 2-8).

Step 2On the System Settings menu, scroll to and select an option, or select Back to return to the Home Screen
or Inspections Settings menu.

OptionDescription

BRIGHTNESS

Set the brightness of the touchscreen.

LANGUAGE

Select the interface language for the microscope.

DATE

Set the date

DATE FORMAT

Set the date format

TIME

Set the time

TIME FORMAT

Set the time format

TOUCHSCREEN CALIBRATION

See “Calibrating the touchscreen” on page 2-20

SYSTEM INFORMATION

Retrieve the following details for the FVAi-2030 microscope:• Device ID• Serial Number• Board Revision• Firmware Version• LCD Revision• Uptime— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-19

Getting started

Managing system settings

Calibrating the touchscreen
If required, you can calibrate the active areas of the touchscreen to improve responsiveness.
Step 1

Access the System Settings menu (see“Managing system settings” on page 2-19), and then select
TOUCHSCREEN CALIBRATION.

The Touchscreen Calibration view, which consists of a white background with a small, red marker in the
top corner, appears.
Step 2

Tap the marker to reveal an additional marker in another area of the touchscreen, and repeat until a redmarker appears in the center.

Step 3

Tap the center red marker to view the results (PASS or FAIL) of the calibration.

Step 4

If calibration is successful, select DONE to return to the System Settings menu. If calibration fails, select
RETRY to repeat the procedure.
Note: If the calibration operation fails repeatedly, power off the microscope to retain the current
calibration settings. You can power the microscope on and continue using it, or you can contact
your VIAVI representative for support.

Step 5

Select Back to return to the Inspection Settings menu.

Step 6

Select another option, or select Back to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-20

Getting started

Connecting and activating a supported USB device

Connecting and activating a supported USB device
You can use with a P5000i Digital Analysis Probe and/or an MP Series Optical Power Meter with the FVAi-2030
microscope in standalone mode by connecting the device to a USB port at the rear of the microscope (see“Features” on page 2-2). When the microscope discovers the device, a tile for the device appears on the Home
screen.
Figure 2-7: FVAi-2030 Home screen example - connected USB devices

Activating a P5000i probe
Note:

For information about operating a P5000i probe, including using the device with the FiberChekPRO
inspection software (via the FVAi-2030 microscope or a direct connection to a computer), see the P5000i
Digital Analysis Probe Getting Started Guide, ZP-PKG-0574, or the FiberChekPRO Automated Fiber
Inspection and Analysis Software User Manual.

Step 1

Select the INSPECT tile for the probe to access Live view.

Step 2

Tap the touchscreen to access the Live view tools for the probe.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-21

Getting started

Connecting and activating a supported USB device

Live View Tool

Description

Home

Return to the Home screen (see Figure 2-7 on page 2-21).

Magnification High/Low

Increase or decrease magnification of the image displayed on the touchscreen.

Microscope Live View

Return to the FVAi-2030 microscope Live view.

Start Test

See “Starting a Pass/Fail test from the FVAi-2030” on page 3-3.
Note: For information about starting a Pass/Fail test with the FiberChekPRO
inspection software, see “Using the FVAi-2030 microscope with FiberChekPRO”
on page 4-1.

CaptureSee “Capturing images” on page 3-5.

Inspection Settings menu
(P5000i probe)

PROFILE: See “Applying a profile and an optical setting” on page 2-10.
Note: FVAi-2030 profiles are not compatible with the P5000i probe. Use theFiberChekPRO inspection software to load compatible profiles to the microscope.
See the P5000i Digital Analysis Probe Getting Started Guide for information.
TIP: See “Applying a profile and an optical setting” on page 2-10.
Note: Ensure that the tip setting you want to use is available on the P5000i probe.
See the P5000i Digital Analysis Probe Getting Started Guide for information.
CREATE NEW GROUP/CLOSE GROUP : GROUP NAME: See “Managing groups” onpage 2-12.
RECALL IMAGE: See“Recalling test results and images” on page 3-7.
USER INFO: See “Managing user information” on page 2-14.
PROBE SETTINGS:• AUTO CENTER• DEVICE BUTTON• SHOW FOCUS METER• DETECT DEFECTS OUTSIDE ZONES• FIBER CENTERING LIMITIMAGE SAVE OPTIONS: See “Managing image-save options” on page 2-18.
SYSTEM SETTINGS: See “Managing system settings” on page 2-19.
OUTPUT LOCAL: See “Navigating the Home screen” on page 2-6.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-22

Getting started

Connecting and activating a supported USB device

Activating an MP Series optical power meter
Note:

•

The information in this section is based on an MP-80 power meter. Details might not apply to a different
MP Series power meter. For information about operating an MP Series power meter, including using thedevice with the FiberChekPRO inspection software (via the FVAi-2030 microscope or a direct connection
to a computer), see the FiberChekPRO Automated Fiber Inspection and Analysis Software User Manual.

Select the tile for the power meter to access the meter tools.

Live View Tool

Description

Home

Return to the Home screen (see Figure 2-7 on page 2-21).

Microscope Live View

Return to the FVAi-2030 microscope Live view.

Power Reading

OPM power reading

Save

Save OPM data

Units

Toggle setting to dB, dBm, or pW unit of measurement
Note: The pW unit of measurement is available only while the ENABLE WATTSDISPLAY setting is enabled.

OPM Settings menu

EDIT WAVELENGTH TABLEREVIEW OPM DATA: Select to access and manage saved OPM data
ENABLE WATTS DISPLAY: Enable availability of the pW unit of measurement.
ENABLE OPM + IMAGE LINK: Enable or disable OPM + IMAGE LINK feature
SYSTEM SETTINGS: See “Managing system settings” on page 2-19.
USER INFO: See “Managing user information” on page 2-14.
OUTPUT LOCAL: See “Navigating the Home screen” on page 2-6.

Wavelength

Select to set wavelength for power measurements.

Reference

Select to obtain a reference measurement.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

2-23

3 Fiber inspection during standalone
operation
This section covers the following information:•“Requirements for Pass/Fail tests” on page 3-2•“Starting a Pass/Fail test from the FVAi-2030” on page 3-3•“Viewing the fiber chamfer” on page 3-4•“Capturing images” on page 3-5•“Recalling test results and images” on page 3-7

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-1

Fiber inspection during standalone operation

Requirements for Pass/Fail tests

Requirements for Pass/Fail tests
Pass/Fail tests inspect fiber connectors against preconfigured pass criteria for the active analysis profile.
Note: For information about viewing the pass criteria for analysis profiles, see the FiberChekPRO AutomatedFiber Inspection and Analysis Software User Manual.
Table 3-1: Before you begin a Pass/Fail test
Ensure that...

See...

A cleaned fiber connector is properly connected to the microscope.

“Mounting fiber adapters and connectors” onpage 2-4

A live image of the fiber connector is displayed on the touchscreen.

“Accessing Live view” on page 2-8

The correct profile and optical setting are applied.

“Applying a profile and an optical setting” onpage 2-10

If required, the group that you want to add saved results to is activeon the microscope.

“Managing groups” on page 2-12

If required, user information you want to appear with saved results
and images is specified.

“Managing user information” on page 2-14

Additional settings are specified as required.

• “Managing microscope settings” on page 2-15
• “Managing image-save options” on page 2-18
• “Managing system settings” on page 2-19

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-2

Fiber inspection during standalone operation

Starting a Pass/Fail test from the FVAi-2030

Starting a Pass/Fail test from the FVAi-2030
Note: Pass/Fail test results saved to the microscope cannot be accessed via FiberChekPRO. If you require use ofFiberChekPRO to generate reports and further manage test results, connect the microscope to acomputer where FiberChekPRO is installed (see “Connecting to FiberChekPRO” on page 4-2) and use
FiberChekPRO to perform the test.
Step 1

Review “Requirements for Pass/Fail tests” on page 3-2.

Step 2In Live view, refer to the Focus-quality meter (if displayed) to determine whether the image quality issuitable for inspection, and apply any adjustments if required.

Step 3

Tap the touchscreen to access the inspection tools (see “Accessing Live view” on page 2-8).

Step 4

Select Start Test.

When the test is completed, analysis details appear on the touchscreen.

Note: For information about zones and overlays, see the FiberChekPRO Automated Fiber Inspection andAnalysis Software User Manual.
Step 5

Tap the touchscreen to display tools for working with the test results and accessing the Inspection
Settings menu.

Step 6

Optionally, select Save to save the results to the microscope.
If a group is active, the saved results are added to it. For information about accessing test results saved tothe microscope, see “Recalling test results and images” on page 3-7.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-3

Fiber inspection during standalone operation

Viewing the fiber chamfer

Viewing the fiber chamfer
In Live view, you can visually analyze the fiber chamfer by using the Chamfer Focus tool to bring the fiber chamfer
into focus (Figure 3-1). The CHAMFER FOCUS OFFSET setting (see “Managing microscope settings” on page 2-15)
lets you control the amount by which the chamfer is brought into focus each time you apply the Chamfer Focus
tool. Setting a lower value applies less focus than a higher value, letting you bring the fiber chamfer into focus
more gradually.
Note:

As Chamfer Focus is applied, the view of the fiber end face becomes less focused.

Figure 3-1: Chamfer focus application (Live view tools hidden)

Step 1

Review “Requirements for Pass/Fail tests” on page 3-2.

Step 2In Live view, tap the touchscreen to access the Live view tools.

Step 3

Select Chamfer Focus to bring the chamfer into focus, repeating until the focus you require is obtained.

Note: Successive applications of the tool after obtaining a focused view of the chamfer will bring thechamfer out of focus.
Step 4Do any of the following:▪ Tap the touchscreen to hide the Live view tools, and then tap again to show them.
▪ Capture an image (see“Capturing images” on page 3-5).
▪ Select Autofocus to reset the focus to the fiber end face.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-4

Fiber inspection during standalone operation

Capturing images

Capturing images
In Live view, you can capture images of the fiber end face and save them to the microscope and recall them forlater analysis. For information about accessing images saved to the microscope, see “Recalling test results andimages” on page 3-7
Note: Captured images saved to the microscope cannot be accessed via FiberChekPRO. If you require use ofFiberChekPRO to manage images, connect the microscope to a computer where FiberChekPRO is
installed and use FiberChekPRO to capture the image (see “Using the FVAi-2030 microscope with
FiberChekPRO” on page 4-1),.
Step 1

Review “Requirements for Pass/Fail tests” on page 3-2.

Step 2In Live view, tap the touchscreen to access the inspection tools (see “Accessing Live view” on page 2-8).

Step 3

Select Capture.

The captured image appears.

Step 4

Tap the touchscreen to display tools for working with the image and accessing the Inspection Settings
menu.

Step 5

Select Save to save the captured image to the microscope.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-5

Fiber inspection during standalone operation

Capturing images

If a group is active, the saved image is added to it, and the group name and fiber ID appears on theimage.

Step 6

Tap the touchscreen to access the tools, and then select Return To Live View.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-6

Fiber inspection during standalone operation

Recalling test results and images

Recalling test results and images
You can access Pass/Fail test results and images saved to the microscope.
Step 1

Do one of the following:▪ On the Home screen, select More, and then select RECALL IMAGE on the Home Screen menu (see“Navigating the Home screen” on page 2-6).
▪ In Live view, select Inspection Settings, and then select RECALL IMAGE on the Inspection Settings
menu (see “Accessing Live view” on page 2-8).

The Recall Image menu lists the groups that saved test results and/or images were added to, as well as
test results and/or images saved but not added to a group.
Step 2

Do one of the following:▪ Individually select the check box for each test result and/or image you want to access, or select the
Select All check box. Test results are indicated by either the Pass or Fail label, and images areindicated by the label Image.

▪ Select a group to access the test results and/or images it contains.

You can individually select test results and/or images in the group, or select Select All.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-7

Fiber inspection during standalone operation

Step 3

Recalling test results and images

Do any of the following:▪ To view selected test results and/or images, go to Step 4.
▪ To delete selected test results and/or images, select Delete, select CONFIRM to confirm the action,and then go to Step 6.

Step 4

Select View.

The first selected test result and/or image is displayed, along with related tools.

Note: Image Information shows the date and time the test results or image was saved, focus metersetting, microscope type, and profile and optical setting applied.
Step 5

Select Back to return to the Recall Image menu.

Step 6

Select another option, or select Back to return to the Home Screen or Inspection Settings menu.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

3-8

4 Using the FVAi-2030 microscope with
FiberChekPRO
This section covers the following information:•“Connecting to FiberChekPRO” on page 4-2•“Performing Pass/Fail tests and generating reports” on page 4-3•“Viewing the fiber chamfer with FiberChekPRO” on page 4-5•“Creating analysis profiles for the FVAi-2030 microscope” on page 4-6

Note: You can download the FiberChekPRO inspection software from http://fcpro.updatemyunit.net/ and install
it on multiple computers at no additional cost. For detailed information about using the software, refer tothe FiberChekPRO Automated Fiber Inspection and Analysis Software User Manual, ZP-SW-00071, which
you can access from the FiberChekPRO Help menu.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-1

Using the FVAi-2030 microscope with FiberChekPRO

Connecting to FiberChekPRO

Connecting to FiberChekPRO
You can use the FVAi-2030 microscope with the FiberChekPRO inspection software via a USB connection to thecomputer where the software is installed (see “Features” on page 2-2). You can then perform Pass/Fail tests, view
test results, generate fiber inspection reports, and manage inspection and microscope settings.
You can also update the microscope firmware or perform a system reset when required. For information, see“Firmware update” on page A-1 and “System reset” on page B-1.
Step 1

Start FiberChekPRO.

Step 2Connect a USB cable to the USB interface on the microscope (see “Features” on page 2-2) and then to
the computer.

Step 3

Power on the microscope (see “Powering the FVAi-2030 microscope on or off” on page 2-3).
The LCD touchscreen on the microscope initializes, and FiberChekPRO automatically detects the
microscope.

Note: While the microscope is connected to the FiberChekPRO inspection software, you can toggle theoutput mode so that you can use either the software or the LCD touchscreen. Select the Home
tile on the touchscreen, access the Home Screen menu, and then select OUTPUT LOCAL on the
Inspection Settings menu. For information, see “Navigating the Home screen” on page 2-6.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-2

Using the FVAi-2030 microscope with FiberChekPRO

Performing Pass/Fail tests and generating reports

Performing Pass/Fail tests and generating reports
FiberChekPRO lets you generate a fiber inspection report using the results of a Pass/Fail test performed with
FiberChekPRO. This report provides a detailed summary of the test performed during an analysis. The report also
includes the date and time that the test was performed, detailed fiber information and, when available,
information such as company name, location, and operator information.
Note: The date and time information on a report reflects the date and time that the Pass/Fail test was
performed. The format used for the date reflects the system settings on the computer where
FiberChekPRO software is installed.
You generate a report by saving the inspection data immediately after performing a Pass/Fail test using
FiberChekPRO. By default, reports are saved to HTML format.
Note: Refer to the FiberChekPRO Automated Fiber Inspection and Analysis Software User Manual for detailed
information about using the tools of the software to analyze and manage test results, as well as
information about specifying the format (HTML or PDF) reports are saved to.
Step 1

Review “Requirements for Pass/Fail tests” on page 3-2.

Step 2Connect the microscope to FiberChekPRO (see “Connecting to FiberChekPRO” on page 4-2).

Step 3In FiberChekPRO, refer to the Focus-quality meter to ensure that live image is suitable for inspection.

Step 4

Click TEST to start the test.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-3

Using the FVAi-2030 microscope with FiberChekPRO

Performing Pass/Fail tests and generating reports

When the test is completed, the results are automatically displayed with overlays.

Step 5

Optionally, click Overlays to hide or show the overlays as needed.

Step 6

Optionally, click Save Report, and then do the following to generate a report:▪ In the Save As dialog, type a name for the report in the Name field, and then click Save to the reportto the default folder on the computer where FiberChekPRO is installed, or navigate to anotherlocation.
The default folder is located at C:\Users\userName\Documents\Viavi\FiberChekPRO\Archives\FVAi2030_deviceID
The report is generated and appears in a separate window.

Step 7

Close the report window when you have finished viewing the information.
You can access the saved report later for viewing or sharing by accessing the folder where the filecontaining the report is located.

Step 8

Select LIVE to return to Live view.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-4

Using the FVAi-2030 microscope with FiberChekPRO

Viewing the fiber chamfer with FiberChekPRO

Viewing the fiber chamfer with FiberChekPRO
You can visually analyze the fiber chamfer by using the Chamfer Focus tool to bring the fiber chamfer into focus(Figure 4-1). The Chamfer Focus Offset setting lets you control the amount by which the chamfer is brought into
focus each time you apply the Chamfer Focus tool. Setting a lower value applies less focus than a higher value,letting you bring the fiber chamfer into focus more gradually.
Notes:• The Chamfer Offset setting is available via the Device Settings option for the microscope. For information
about accessing the Device Settings option, see the FiberChekPRO Automated Fiber Inspection and Analysis
Software User Manual.
• As Chamfer Focus is applied, the view of the fiber end face becomes less focused.
Figure 4-1: Chamfer Focus application.

Step 1

Review “Requirements for Pass/Fail tests” on page 3-2.

Step 2Connect FiberChekPRO to the microscope (see “Connecting to FiberChekPRO” on page 4-2).

Step 3In FiberChekPRO, refer to the Focus-quality meter to ensure that the live image is suitable for inspection.

Step 4

Select Chamfer Focus on the Image Controls toolbar to bring the chamfer into focus, repeating until the
focus you require is obtained.
Note: Successive applications of the tool after obtaining a focused view of the chamfer will bring thechamfer out of focus.

Step 5

Optionally, do any of the following:▪ Select Save, and then navigate to the folder when you want to save the image.
Note: Refer to the FiberChekPRO Automated Fiber Inspection and Analysis Software User Manual for
detailed information about using the tools of the software to save and manage images.
▪ Select Auto Focus to reset the focus to the fiber end face.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-5

Using the FVAi-2030 microscope with FiberChekPRO

Creating analysis profiles for the FVAi-2030 microscope

Creating analysis profiles for the FVAi-2030 microscope
In addition to providing factory-installed profiles specific to the FVAi-2030 microscope, FiberChekPRO provides
tools to help you create profiles to meet your fiber-inspection requirements.
You can create either a new profile or a customized copy of an existing FVAi-2030 profile.
Note: For detailed information about using FiberChekPRO to manage analysis profiles, see the FiberChekPRO
Automated Fiber Inspection and Analysis Software User Manual.
Step 1Connect the microscope to FiberChekPRO (see “Connecting to FiberChekPRO” on page 4-2).

Step 2On the Setup menu, select the microscope.

Step 3On the Setup page for the microscope, select Analysis Profiles in the navigation pane to access the lists(Available and On Device) of analysis profiles.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-6

Using the FVAi-2030 microscope with FiberChekPRO

Step 4Creating analysis profiles for the FVAi-2030 microscope

Do one of the following:▪ To create a new FVAi-2030 profile, select New to access the Edit Profile dialog.

▪ To create a customized copy of an existing FVAi-2030 profile, select an FVAi-2030 profile in either list,
and then select Copy to access the Edit Profile: Copy of dialog.

Note: Ensure that you select an FVAi-2030 profile to copy. Profiles intended for use with other devices
should not be modified for use with the microscope.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-7

Using the FVAi-2030 microscope with FiberChekPRO

Step 5Creating analysis profiles for the FVAi-2030 microscope

On the Setup page, specify or modify general settings for the profile.

Parameter

Description

Profile Name

Unique name of the profile.
Note: FVAi-2030 analysis profiles define settings exclusive to the microscope (see theparameter FVAi-2030 in this table). VIAVI recommends including FVAi-2030 in the nameof the profile to help identify it as an FVAi-2030-specific profile.

Number of Zones

Number of zones of interest. Refer to the Zone Preview view in the dialog.
Note: Zones appear as series of concentric circles on the view of the fiber end face, startingwith the smallest zone centered on the core (see “Performing Pass/Fail tests andgenerating reports” on page 4-3). Failure criteria is defined for each zone.

Fiber TypeType of fiber being tested.

Cladding Diameter (µm)

Fiber cladding diameter (Default = 125 µm).

Core Diameter (µm)

Fiber core diameter.
Note: FiberChekPRO identifies the core with an overlay of the core diameter in addition to
the specified zones (see “Performing Pass/Fail tests and generating reports” on page 4-3).

Scratches

Detection parameters for scratches:• Find Scratches: Enable to detect any scratch.
• Find dark scratches: Enable to detect scratches that are dark.
Note: Enabling this feature will significantly impact the amount of time required tocomplete a Pass/Fail test.
• Find no more than: Set a limit for the number of scratches to be detected.
Note: VIAVI recommends setting a high value (e.g., 25) to speed up analysis of severely
scratched fiber end faces.

FVAi-2030

Settings for use with the FVAi-2030 microscope:• Ferrule Diameter (µm): Ferrule diameter of fiber being tested. (Required setting.)• Automated polish area detection: Enable to automate detection of polish area.

Profile Notes

Information about the profile.

Show/Hide Advanced Setup
Options

Access or hide setup options:• Sensitivity: Parameterize for greater or lesser sensitivity (on a scale of 0 to 10), affecting
what defects will be found (i.e., cladding dirt, cladding pits/chips, ferrule dirt, and/or
scratches.
Low settings enable detection of easily visible scratches and defects, and define them asexisting only in the zones where these are clearly present. High settings enabledetection of extremely faint scratches and defects that are otherwise difficult to detect
reliably.
Note: High settings are intended for use only in post-polish applications for processcontrol purposes, and are not recommended for connectors that are already in the field.
• Overall Inspection Criteria: Define specific criteria relating to epoxy gap and core
saturation.
• Add Password Protection: Enable password protection on the profile.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-8

Using the FVAi-2030 microscope with FiberChekPRO

Step 6Creating analysis profiles for the FVAi-2030 microscope

For each available zone, select the tab for the zone and specify settings as required:▪ Name, Inner Diameter, Outer Diameter, and Outer Diameter Color for the zone.
▪ Scratches: Specific acceptance criteria for scratches.
▪ Defects: Specific acceptance criteria for defects, including any non-scratch defect on the connector(dirt, debris, pits, chips). Specific criteria can be defined for both Individual and Combined defects.
Select Enable Advanced Criteria to define additional options to fine-tune acceptance criteria that
address specific requirements.
▪ Exclusion Zone: Inner and outer radius in which defects and scratches are not counted toward theacceptance criteria.

Step 7

Select OK to close the dialog for the profile.
A new FVAi-2030 profile is automatically added to the On Device list. An FVAi-2030 profile created using
a copy of an existing profile is automatically added to the same list from which the original profile wasselected (Step 4).

Step 8

Select OK in the navigation pane to return to the Inspection window.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

4-9

Appendix A Firmware update
FiberChekPRO facilitates firmware management by identifying available firmware update versions for each device
it supports, including the FVAi-2030 microscope.
Note: Firmware for the FVAi-2030 microscope is specific to the FVAi-2030 model and is not supported by other
FV Series microscope models.
Performing a firmware update
Step 1Connect the microscope to FiberChekPRO (see “Connecting to FiberChekPRO” on page 4-2).
If the microscope is running an earlier firmware version and FiberChekPRO is set up to automaticallycheck for updates at startup, the New Firmware dialog appears in the FiberChekPRO application
window, prompting you to download the updated firmware to your computer.

Step 2. Do the following:i. In the New Firmware dialog, click Download Now.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

A-1

Appendix A

Firmware update

ii. In the FiberChekPRO Setup dialog, click Download for the microscope model.

iii. When the download is completed, click OK in the FiberChekPRO Setup navigation pane to return to
the FiberChekPRO application window.
Step 3. On the Setup menu, select the microscope.

Step 4. On the Setup page for the microscope, select the Firmware option in the navigation pane, and then click
Update Firmware.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

A-2

Appendix A

Firmware update

Step 5. Click Update to start the update.

Step 6. When the update is successfully completed, click OK to reboot the microscope.

After the microscope reboots, it automatically reconnects to FiberChekPRO.
Step 7. Access the Firmware option on the Setup page for the microscope (see Step 3 and Step 4), and note thefirmware version.
Step 8. Select OK in the navigation pane to return to the Inspection window.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

A-3

Appendix B

System reset

When required, you can perform a system reset to return the firmware running on an FVAi-2030 microscope toan earlier version. From that version, you can update to the firmware version you require. This facilitates
maintaining firmware-version stability and uniformity across a fleet of deployed microscopes.
Note: If you must return to an earlier version of the firmware, you must first perform a system reset. You canthen update to a firmware version later than the version defined by the system-reset threshold.
Important:• VIAVI strongly recommends that you refer to the document Standardizing Device Firmware for information
about the system reset process, including limitations, before you perform this procedure.
• A system reset deletes all test results, custom analysis profiles and optical settings available on the
microscope, and restores all factory-set options. Therefore, before you start a system reset, ensure that all
required custom profiles and optical settings are available on FiberChekPRO. For information, seeFiberChekPRO Automated Fiber Inspection and Analysis Software User Manual.
Performing a system reset
Step 1Connect the microscope to FiberChekPRO (see “Connecting to FiberChekPRO” on page 4-2).

Step 2On the Setup menu, select the microscope.

Step 3On the Setup page for the microscope, select Firmware in the navigation pane, and then click System
Reset.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

B-1

Appendix B

System reset

Note: The System Reset button is available only while the firmware version installed on the microscopeis either the same version defined by the system-reset threshold or a later version.
Step 4

Click Next to either download the system reset package or proceed to the next dialog.
Note: The system reset package is downloaded only during a first-time system reset.

A dialog containing warning information appears.

Step 5

Review the information, and then click Reset to start the system reset.

The system reset can take several minutes to complete.
Step 6

When the system reset is successfully completed, click OK to reboot the microscope.

After the microscope reboots, it automatically reconnects to FiberChekPRO.
— End —

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

B-2

Appendix C Specifications
Table C-1: FVAi-2030 Benchtop Microscope specifications
Parameter

Specification

Auto-focus enabled

Yes

Particle Size Detection< 10 µm

Camera Sensor
Built-in LCD Screen Size, Resolution
Light Source, Technique
Adapter TypeUSB Ports
External Display Port, Resolution
Dimensions (H x W x D)

2560 x 1920, 1/2.5-in CMOS, 5 megapixels
High-contrast 3.5-in color LCD with touchscreen functionality
Live Image (pixels): 320 x 240 x 8 bit gray, 10 fps
Blue LED (100,000+ hours life)
CoaxialFMAX SL Series
Host (x1): USB Type B
Device (x2): USB Type A
VGA, 640 x 480
214 x 135 x 211 mm (8.4 x 5.3 x 8.3 in) See Figure C-1.

Weight3.1 kg (6.8 lb)

Power Supply

12V, 2A with interchangeable wall plug for EU, UK, US, and AU

Certification

CE, EN/IEC 61326

Figure C-1: FVAi-2030 Benchtop Microscope dimensions (mm)FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

C-1

Appendix D Document revision history
Table 1: FVAi-2030 Benchtop Microscope User Guide, 22112369-339
Revision

Date

Details

R002

November 2023

• Added the section “Safety instructions”• Updated the section “Accessing Live view”

R001

November 2022

• Updated frontmatter content.
• Updated VIAVI address on back cover.

R000

December 2021

This document is released.

FVAi-2030 Benchtop Microscope User Guide, 22112369-339 R002, Standard

D-1

Multiple Application Platform

FVAi-2030 Benchtop Microscope User Guide
22112369-339 R002, Standard
November 2023
EnglishVIAVI Solutions Inc.
1445 South Spectrum Blvd., Suite 102
Chandler, AZ, 85286
USA

North America:
All Other Regions:(Toll Free) 1-844-GO-VIAVI / 1-844-468-4284
www.viavisolutions.com/contacts