PT0-1582

ポータブル酸素モニター

OX-08
取扱説明書
（PT0-158）〒174-8744 東京都板橋区小豆沢 2-7-6
ホームページ http://www.rikenkeiki.co.jp/

目次
1. 製品のアウトライン .............................................................................................................................................................................................................................. 4
はじめに ............................................................................................................................................................................................................................................... 4
使用目的 .............................................................................................................................................................................................................................................. 4
危険、警告、注意、注記の定義 ............................................................................................................................................................................................................. 5
2. 安全上、大切なお知らせ...................................................................................................................................................................................................................... 62-1. 危険事項 ...................................................................................................................................................................................................................................... 62-2. 警告事項 ...................................................................................................................................................................................................................................... 82-3. 注意事項 ...................................................................................................................................................................................................................................... 9
3. 製品の構成 ........................................................................................................................................................................................................................................ 113-1. 本体および付属品 ....................................................................................................................................................................................................................... 113-2. 各部の名称と働き ...................................................................................................................................................................................................................... 133-3. 電池の装着方法 ......................................................................................................................................................................................................................... 153-4. ベルトクリップの装着方法 ........................................................................................................................................................................................................... 173-5. リモートケーブルの脱着方法 ...................................................................................................................................................................................................... 183-6. リモートケーブルの巻き取り方法 ................................................................................................................................................................................................. 213-7. ネックストラップ及び解け防止バンドの取り付け方法.................................................................................................................................................................... 233-8. 延長ケーブル（オプション）の取り付け方法 .................................................................................................................................................................................. 25
4. 使用方法........................................................................................................................................................................................................................................... 27
4-1. ご使用にあたって ....................................................................................................................................................................................................................... 274-2. 始動準備 .................................................................................................................................................................................................................................... 274-3. 電源を入れる ............................................................................................................................................................................................................................. 284-4. 基本動作フロー .......................................................................................................................................................................................................................... 304-5. ディスプレイモード ...................................................................................................................................................................................................................... 314-6. エア校正する .............................................................................................................................................................................................................................. 33
4-7. 検知する .................................................................................................................................................................................................................................... 35
4-8. 電源を切る ................................................................................................................................................................................................................................. 36
5. 警報機能........................................................................................................................................................................................................................................... 37
5-1. ガス警報動作 ............................................................................................................................................................................................................................. 37
5-2. 故障警報動作 ............................................................................................................................................................................................................................ 39
5-3. 警報の解除 ................................................................................................................................................................................................................................ 40
6. 保守点検........................................................................................................................................................................................................................................... 41
6-1. 点検の頻度と点検項目 ............................................................................................................................................................................................................... 416-2. 清掃方法 .................................................................................................................................................................................................................................... 42
6-3. 各部品の交換 ............................................................................................................................................................................................................................ 43-2-6-4. フィルタ交換 ............................................................................................................................................................................................................................... 44
7. 保管および廃棄について ................................................................................................................................................................................................................... 46
7-1. 保管または長期間使用しない場合の処置 ................................................................................................................................................................................... 46
7-2. 製品の廃棄 ................................................................................................................................................................................................................................ 47
8. トラブルシューティング ....................................................................................................................................................................................................................... 48
8-1. 機器の異常 ................................................................................................................................................................................................................................ 48
9. 製品仕様........................................................................................................................................................................................................................................... 50
9-1. 仕様 ........................................................................................................................................................................................................................................... 50
10. 付録 ................................................................................................................................................................................................................................................ 5210-1. 用語の定義 .............................................................................................................................................................................................................................. 5210-2. 検知原理（ガルバニ電池式） ..................................................................................................................................................................................................... 53-3-

1. 製品のアウトライン

はじめに

1製品のアウトライン
はじめに
この度は、ポータブル酸素モニター OX-08（以降「本器」）をお買い上げいただきありがとうございます。お買い求めの製品型番と本取扱説明書に記載の仕様を照合し、ご確認をお願いします。
本書は本器の取扱方法と仕様を説明したものです。本器を正しく使用していただくために必要な事項が記載されています。初めて使用される方はもちろん、既に使用されたことのある方も、知識や経験を再確認する上で、よくお読みいただき、内容を理解した上で使用してください。
なお、製品改良のために、この説明書の内容を将来予告なしに変更することがあります。また、この説明書の全部または一部を無断で複写または転載することを禁じます。
保証期間の内外を問わず本器を使用することによって生じたいかなる事故および損害の補償はいたしません。
保証書に記載されている保証規定を必ずご確認ください。

使用目的
本器は、大気中の酸素を検知し、ガス警報を発し、酸素欠乏による事故を防止するための小型ガスモニターです。

-4-

1. 製品のアウトライン

危険、警告、注意、注記の定義

危険、警告、注意、注記の定義
本取扱説明書では、安全かつ効果的な作業が行えるように、次の見出しを使用しています。

危険取り扱いを誤った場合、「人命、人体または物に重大な被害を及ぼすことが想定される」ということを意味します。

警告

取り扱いを誤った場合、「身体または物に重大な被害を及ぼすことが想定される」ということを意味します。

注意

取り扱いを誤った場合、「身体または物に軽微な被害を及ぼすことが想定される」ということを意味します。

注記

取り扱い上のアドバイスを意味します。

-5-3. 製品の構成

3-1. 本体および付属品

3

製品の構成
3-1. 本体および付属品梱包箱を開けて、本器と付属品を確認します。
万一、足りないものがありましたら、販売店または最寄りの弊社営業所までお問い合わせください。

本体
本器の各部の名称と働きおよび LCD 表示は、「3-2. 各部の名称と働き」（P.13）を参照してください。

- 11 -

3. 製品の構成

3-1. 本体および付属品

標準付属品
リモートケーブル（5m）：
1本

•
••

プロテクトカバー：1個

単 3 アルカリ乾電池：
2 本（実装）

ベルトクリップ：
1 個（実装）

ネックストラップ：1本

取扱説明書：1 冊
製品保証書：1 枚
センサ保証書：1 枚

オプション（別売品）
延長ケーブル（5m,15m,25m）

収納ケース（アルミトランクケース）- 12 -LCD 保護フィルム（5 枚入り）

解け防止用バンド：1本3. 製品の構成

3-3. 電池の装着方法

3-3. 電池の装着方法
初めて使用する場合や電池残量が少ない場合は、以下の要領で新品の単 3 形アルカリ乾電池 2 本を装着、または交換してください。

1

本器の電源が切れていることを確認する
電源が入っている場合は、いったん切ってください。

2

コインなどで電池蓋固定ネジを反時計方向に回して緩め、電池蓋を開ける

3

古い電池を取り出し、電池の極性に注意して新しい電池を入れる
本器背面に刻印された電池の極性に従って、電池を入れてください。

4

電池蓋を閉め、電池蓋固定ネジを時計方向に回してロックする- 15 -

3. 製品の構成

3-5. リモートケーブルの脱着方法

3-5. リモートケーブルの脱着方法
ここでは、リモートケーブルの取り付け/取り外し方法を説明します。

ケーブルの取り外し
1 電源が切れていることを確認する
2 ロックリングを回し、ネジを外す

3 ロックリングを上方に引き抜く- 18 -

3. 製品の構成

3-5. リモートケーブルの脱着方法

ケーブルの取り付け
1 電源が切れていることを確認する
2 ロックリング上部の▼マークを本器正面に向け、接続部に差し込む

3 ロックリングを回してネジを締め込む- 19 -

4. 使用方法

4-3. 電源を入れる

注記• 本器の使用温度範囲は、－20～＋50℃ですが、急激に温度が変化するような場所で測定を行うと、温度差の影響により指示が変動することがあります。
• 約 20 秒間、ボタン操作が何も行われない場合は、自動的に検知モードへ戻ります。
• AIR 、POWER/MODE のいずれかのボタンを押すと、LCD バックライトが点灯します。LCD バックライトは、約 30 秒間無操作で消灯します。
また、警報発報時には自動的に LCD バックライトが点灯します。
• 電池電圧表示による電池残量の目安は、以下の通りです。
電池電圧表示電池残量表示

使用可能時間 目安

2.5V 以上

6,000～20,000 時間

2.3～2.4V

4,000～6,000 時間

2.2～2.3V

2,000～4,000 時間

2.1～2.2V（点滅）

0～2,000 時間- 29 -

4. 使用方法

4-5. ディスプレイモード
フルスケール値、第一警報点設定値（WARNING）、第二警報点設定値（ALARM）が表示されているときには、AIR ボタンと POWER/MODE ボタンを同時に押すと、警報テストを行うことができます。

フルスケール値、第一警報点設定値（WARNING）、第二警報点設定値（ALARM）が表示されているときに、POWER/MODE ボタンを押すと、検知モードに戻ります。

- 32 -

4. 使用方法

4-6. エア校正する

注記• エア校正不良となった場合は、LCD に「AIR CAL～FAIL」（校正不良）と表示されます。警報の解除は、POWER/MODE ボタンを押してください。警報が解除されると、LCD にはエア校正前の値が表示されます。

- 34 -

6. 保守点検

6-4. フィルタ交換

6-4. フィルタ交換
フィルタは消耗部品です。汚れの具合を見て定期的に交換してください。
フィルタを交換するときは、必ず電源を切ってから行ってください。

1 電源が切れていることを確認する
2 センサケースを回し、ケースを外す

3 パッキンとフィルタを外し、新しいものと交換する

4 パッキンをセンサに被せ、ケースを取り付ける- 44 -

改廃履歴版

修 正発行日0初版

2017/9/20

1

P26 保護フィルムに関する警告追記

2017/12/18

2

P12 オプション(別売品)追記、P25 3-8.延長ケーブル（オプション）の取り付け方法追記

2019/5/7