EA01051D/06/A2/03.18
71388393

ProductsSolutions

Services

Installation Instructions
Safety Instructions for SpareParts

EN Installation Instructions: EA01051D →  3
DE Einbauanleitung: EA01051D →  7

EA01051D

Safety instructions for spare parts
Promag 10, 40, 50, 51, 53, 55, 100, 200, Promag 400 Index B, Promag 400 Index C,Promag 800, Promass 80, 83, 84, 100, 200, CNGmass, LNGmass, LPGmass,Cubemass CNG, Cubemass DCI, Prosonic Flow 91, 92, 93, 200, t-mass 65, 200, 150,
Prowirl 72, 73, 200
Table of contents
1

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

Endress+Hauser

3

EA01051D

1

Designated use

The spare part set and the Installation Instruction are used to replace a faulty unit with afunctioning unit of the same type. Use genuine parts from Endress+Hauser only.
Only original spare part sets, intended by Endress+Hauser for the measuring device, must beused.
The verification has to be done via W@M Device Viewer; this procedure is explained below.
For some devices there is an overview of spare part sets inside the device. If the sparepart set is listed there, the verification is not required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

Order code

4.

2

=Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (Ser. No.), search for productdata and click on "spare parts".

3.

List of all spare parts sets for the device.

4.

Determine the order number of the spare part set.

The spare part set can only be used if the order numberof the spare part set corresponds to theorder number on the spare part list.

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

4

Measuring device approval

Personnel authorized to perform repairs 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Endress+Hauser

EA01051D

3

Safety instructions• Check whether the spare part matches the identification label on the measuring device, as
explained on the first page.
• The spare parts set and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Use genuine parts from Endress+Hauser only.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• Requirements with regard to specialized technical staff for the mounting, electricalinstallation, commissioning, maintenance and repair of the measuring devices:– trained in instrument safety.
– familiar with the individual operation conditions of the devices.
– for Ex-certified measuring devices: also trained in explosion protection.
• The measuring device is energized. Danger: Risk of electric shock! Open the measuring
device in a de-energized state only.
• In the case of Ex-certified measuring devices: Only open in a de-energized state (once adelay of 10 minutes has elapsed after switching off the power supply) or in environments
which do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: After repair recommission in accordance with Operating
Instructions. Document the repair procedure.
• Before removing the device: set the process in a safe condition and purge the pipe ofdangerous materials.
• Hot surfaces! Risk of injury! Before commencing work, allow the system and measuring
device to cool down to a touchable temperature.
• In the case of measuring devices in custody transfer, the custody transfer status no longer
applies once the lead seal has been removed.
• Follow the Operating Instructions for the device.
• Risk of damaging electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics cover, there is a risk of electric shock as shock protection isremoved!
Switch off the measuring device before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open housing for a brief period. Avoid the penetration of foreign bodies, moisture orcontaminants.
• Replace defective seal/gaskets with genuine parts from Endress+Hauser only.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the cover for the electronics and connection compartments) must belubricated. Use an acid-free, non-hardening grease if an abrasion resistant dry lubricant isnon-existent.

Endress+Hauser

5

EA01051D• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service connector:– Do not connect in potentially explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the OperatingInstructions.
If you have any questions, contact your Endress+Hauser service organization.

6Endress+Hauser

EA01051D

Sicherheitshinweise für Ersatzteile
Promag 10, 40, 50, 51, 53, 55, 100, 200, Promag 400 Index B, Promag 400 Index C,
Promag 800, Promass 80, 83, 84, 100, 200, CNGmass, LNGmass, LPGmass,Cubemass CNG, Cubemass DCI, Prosonic Flow 91, 92, 93, 200, t-mass 65, 200, 150,
Prowirl 72, 73, 200
Inhaltsverzeichnis
1Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

2Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

3Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

Endress+Hauser

7

EA01051D

1

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

2

Order code

=Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für dieSicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

8Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3Mit Zulassung (z.B. IECEx)

1, 2, 3Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Endress+Hauser

EA01051D

3Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur imspannungslosen Zustand öffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosem Zustand (nach Berücksichtigung
einer Wartezeit von 10 Minuten nach Abschalten der Energiezufuhr) oder in Umgebungenöffnen, die keine explosionsfähige Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist dergeeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.

Endress+Hauser

9EA01051D• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

10Endress+Hauser

*71388393*
71388393

www.addresses.endress.com