Products

EA01135D/06/A2/02.16
71311436

Solutions

Services

Installation Instructions

Repairkit cover remote version stainless steel
Promag 10, 50, 53, 55 H

Instruction is valid for the following spare part kits:
Order number

Original Spare part kit

71309084

Cover repair kit for G04

NOTICE‣ The order number of the spare part set (on the packaging label) can differ from the product number (on the label directly on thespare part)!

‣ The order number of the relevant spare part set can be established by entering the material number of the spare part in the spareparts finder.

‣ We recommend that the Installation Instructions be kept with the packaging at all times.
Appropriate Use
The spare parts set and Installation Instructions are used to replace a faulty unit with a functioning unit of the same type. Use genuine
parts from Endress+Hauser only.
Only original spare parts supplied by Endress+Hauser shall be used with the measuring device.
The verification has to be done via W@M Device Viewer, this procedure is explained below.
For some devices there is an overview of the correct spare parts inside the connection compartment cover.
If the spare part is listed there, the verification is no longer required.

1.

www.endress.com/deviceviewer

1.

www.endress.com/deviceviewer

2.

Enter the serial number (Ser. No.), search for product data and clickon „spare parts“.

2.

Ser. No.:12345

3.

3.

Display of all suitable spare parts.

4.

Check the order number of the spare part.

4.

=?

Does the order number of spare part correspond to the order number inthe spare part list?
= Yes, the spare part can be used.
A0022408= NO, the spare part may not be used.

Repairkit cover remote version stainless steel

Authorized personnel
NOTICEThe person who carries out the repair is responsible for safety during the work, the quality of work completed and safety of the device
after repair.
Approval of the measuring device

Group of persons authorized to carry out repairs

without approval

1, 2, 3

with approval (for Ex. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 = Trained customer technician
2 = Service technician authorized by Endress+Hauser
3 = Endress+Hauser (send measuring device back to manufacturer)

Safety instructions• Check whether the spare part matches the identification label on
the measuring device, as explained on the first page.
• The spare parts set and Installation Instructions are used toreplace a faulty unit with a functioning unit of the same type.
Use genuine parts from Endress+Hauser only.
• In the case of Ex-certified measuring devices: Only open ina de-energized state (once a delay of 10 minutes has elapsed
after switching off the power supply) or in environments which
do not have a potentially explosive atmosphere.
• The measuring device is energized. Danger: Risk of electric
shock! Open the measuring device in a de-energized state only.
• Before removing the device: set the process in a safe conditionand purge the pipe of dangerous materials.
• Hot surfaces! Risk of injury!
Before commencing work, allow the system and measuring
device to cool down to a touchable temperature.
• In the case of measuring devices in custody transfer, the custodytransfer status no longer applies once the lead seal has beenremoved.
• Comply with national regulations governing mounting, electricalinstallation, commissioning, maintenance and repair
procedures.
• Requirements with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenanceand repair of the measuring devices:– trained in instrument safety– familiar with the individual operation conditions of the devices– for Ex-certified measuring devices: also trained in
explosion protection• Follow the Operating Instructions for the device.
• Risk of damaging electronic components!
Ensure you have a working environment protected from electrostatic discharge.

2• After removing the electronics cover, there is a risk of electric
shock as shock protection is removed! Switch off the measuring
device before removing internal covers.
• Modifications to the measuring device are not permitted.
• In the case of measuring devices in safety-related applications inaccordance with IEC 61508 or IEC 61511:
After repair recommission in accordance with Operating
Instructions. Document the repair procedure.
• Only open housing for a brief period. Avoid the penetration offoreign bodies, moisture or contaminants.
• Caution! When replacing amplifier boards, I/O boards or submodules, ensure compatibility with existing software. The process
for reading out the software revision number is described in the
Operating Instructions (device functions).
If the software of the board is not compatible, it should be updated using an operating software package (e.g. Field Care). In theevent of functional changes, notify the plant owner/operator.
• Replace defective seal/gaskets with genuine parts fromEndress+Hauser only.
• If threads are damaged or defective, the measuring device must
be repaired.
• Threads (e.g. of the cover for the electronics and connection compartments) must be lubricated. Use an acid-free, non-hardening
grease if an abrasion resistant dry lubricant is non-existent.
• If spacing is reduced or the dielectric strength of the measuring
device cannot be guaranteed during repair work, perform a test
on completion of the work (e.g. high-voltage test in accordancewith the manufacturer's instructions).
• Service connector:– do not connect in potentially explosive atmospheres.
– only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning thedevice outlined in the Operating Instructions.
• If you have any questions, contact your Endess+Hauser service
organization.

Endress+HauserRepairkit cover remote version stainless steel

Tool list

8 mm

Safety symbols
DANGER!
This symbol alerts you to a dangerous situation. Failure to avoid this situation will result in serious or fatal injury.

DANGER
A0011189-EN

WARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result in serious or fatal injury.

WARNING
A0011190-EN

CAUTION!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result in minor or medium injury.

CAUTION
A0011191-EN

NOTICE!
This symbol contains information on procedures and other facts which do not result in personal injury.

NOTICE
A0011192-EN

Symbols for certain types of information
Permitted
Procedures, processes or actions that are permitted.
A0011182

Forbidden
Procedures, processes or actions that are forbidden.
A0011184

A0011193

1.

2.

Endress+Hauser

3.

Tip
Indicates additional information.
Series of steps

3Repairkit cover remote version stainless steel

Replacing cover
NOTICE

Open the cover.

2.

The original cover is removed.

3.

Make sure that the defective cover including seal is scrapped (see graphic 3 below). No re-use!

4.

Afix the label „Use exclusively Cover Revision 1“ into the housing to be repaired (see graphic 4 below).

5.

The new cover is ready for mounting.

6.

Close the new cover and tighten the screws with at least 3,5 Nm (2.6 lbf ft).
3.
21319480-0000A

yelge

br

88 mm
mm

gn

2.
whws

1.

whws

5! Cable.= dN.cisolate
beN.c.

7brw
w hsc.

N.

5
37

20A

00

-080

c.

4N.

shleabC dte. = la.c o
N e isb794

31!
EP1ils41Co

42gn

0A0-000

31948

c.

N.

ye
ge l

N.c.

37r ernso
Se ehmDufn
EP to ssas Meeldshiyelge

2
1N.c.

42 41

5 7 4 37!

gn

4N.c.

N.c.

N.c.

br

Coils

EPD

D

ie
lds
42

to M

oilsr

r eso hm
en e
S ufn
saes

C

41

N.c. = Cable shields toSensor
be isolated
Messaufnehmer

A0012879

A0027065

A0027205

319480-0000A

2gn

yelge

br

5.
whws

4.

1

1.

seU
elysivclu
exerov
Cnioisev
R1N.c.

N.c.

N.c.

5 7 4 37ww hs

br!

c.
N.

yege lgnc.
N.

50A
00
-080
94
31!
74c.
N.

shle
abC dte. = la.c o
N e isb1

237sld

ieDPE
soil

C

1 rn eio or 4m
isns 0e8heve 9
R S 130ufn
7 sa
es

41

A0028643

42to MUse exclusively Cover Revision 1

EPD

42 41
Coils

N.c. = Cable shields to Revision 1
Sensorbe isolated
71309084
Messaufnehmer
A0028644

6.

88 mm
mm>3,5 Nm(>2.6 lbf ft)

A0028646

4Endress+Hauser

Products

EA01135D/06/A2/02.16
71311436

Solutions

Services

Einbauanleitung

Reparaturset Deckel Getrenntausführung Edelstahl
Promag 10, 50, 53, 55 H

Die Einbauanleitung ist für folgende Ersatzteilsets gültig:
Bestellnummer

Original Ersatzteilset

71309084

Deckel Reparaturkit zu G04

HINWEIS‣ Die Bestellnummer des Ersatzteilsets (auf dem Produktaufkleber der Verpackung) kann sich von der Produktionsnummer (aufdem Aufkleber direkt auf dem Ersatzteil) unterscheiden!

‣ Durch Eingabe der Produktionsnummer des Ersatzteiles im Ersatzteilfindetool kann die Bestellnummer des ensprechenden
Ersatzteilsets ermittelt werden.

‣ Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.
Bestimmungsgemäße Verwendung
Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen eine funktionierende Einheit des gleichen Typs zu
ersetzen. Es dürfen nur Originalteile von Endress+Hauser verwendet werden.
Grundsätzlich dürfen nur Ersatzteilsets verwendet werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise ist nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Anschlussraumdeckel eine Übersicht der passenden Ersatzteile.
Ist dieses Ersatzteil dort aufgelistet, entfällt die Überprüfung.

1.

www.endress.com/deviceviewer

2.

Ser. No.:12345

3.

4.

=?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten suchen und auf"Ersatzteile" klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Bestellnummer des Ersatzteilsets ermitteln.

Stimmt die Bestellnummer auf dem Produktaufkleber mit der Bestellnummer in der Ersatzteilliste überein?
= JA, das Ersatzteilset darf für das Messgerät verwendet werden.
A0022408= NEIN, das Ersatzteilset darf für das Messgerät nicht verwendetwerden.

Reparaturset Deckel Getrenntausführung Edelstahl

Reparaturberechtigte Personen
HINWEISDie Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die Sicherheit während der Arbeiten, die Qualität derAusführung und die Sicherheit des Geräts nach der Reparatur.
Zulassung des Messgeräts

Reparaturberechtigter Personenkreis

ohne Zulassung

1, 2, 3

mit Zulassung (z.B. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 = Ausgebildete Fachkraft des Kunden
2 = Von Endress+Hauser autorisierter Servietechniker
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf
dem Messgerät passt, wie auf der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte
Einheit gegen eine funktionierende Einheit des gleichen Typs zuersetzen. Nur Originalteile von Endress+Hauser verwenden.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosemZustand (nach Berücksichtigung einer Wartezeit von
10 Minuten nach Abschalten der Energiezufuhr) oder in
Umgebungen öffnen, die keine explosionsfähige Atmosphäre
enthalten.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag.
Messgerät nur im spannungslosen Zustand öffnen.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen
und Leitung von gefährlichen Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen!
Vor Arbeitsbeginn: Anlage und Messgerät auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach
Entfernen der Plombe ist der geeichte Zustand aufgehoben.
• Nationale Vorschriften bezüglich der Montage, elektrischen
Installation, Inbetriebnahme, Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage,elektrische Installation, Inbetriebnahme, Wartung undReparatur der Messgeräte müssen erfüllt sein:–In Gerätesicherheit ausgebildet–Mit den jeweiligen Einsatzbedingungen der Geräte vertraut–Bei Ex-zertifizierten Messgeräten zusätzlich im
Explosionsschutz ausgebildet• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile!
Eine ESD-geschützte Arbeitsumgebung herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromchlaggefahr
durch aufgehobenen Beruhrungsschutz!
Messgerat ausschalten, bevor interne Abdeckungenentfernt werden2• Änderungen am Messgerät sind nicht zulässig.
• Bei Messgeräten in sicherheitstechnischen Applikationen
gemäß IEC 61508 bzw. IEC 61511:
Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung
durchführen. Reparatur dokumentieren.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern,
Feuchtigkeit oder Verunreinigung vermeiden.
• Achtung! Beim Auswechseln von Messverstärker-, I/O-Platineoder Submodulen: Kompatibilität mit der vorhandenen Software
sicherstellen. Der Auslesevorgang der Software
Revisionsnummer ist in der Betriebsanleitung (Gerätefunktionen) beschrieben. Wenn die Software der Platine nicht
kompatibel ist, muss mit Hilfe einer Bediensoftware (z. B. FieldCare) ein Update durchgeführt werden. Bei funktionalenÄnderungen Anlagenbetreiber informieren.
• Defekte Dichtungen nur durch Original-Dichtungen vonEndress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z. B. von Elektronikraum-und Anschlussraumdeckel)
müssen geschmiert sein. Säurefreies, nicht härtendes Fettverwenden, sofern keine abriebfeste Trockenschmierungvorhanden ist.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die
Spannungsfestigkeit des Messgeräts nicht sichergestellt werden
kann: Prüfung nach Abschluss der Arbeiten durchführen
(z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– nicht in explosionsfähiger Atmosphäre anschließen.
– nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum
Transport und zur Rücksendung beachten.
• Bei Fragen kontaktieren Sie bitte Ihre zuständige
Endress+Hauser Serviceorganisation.

Endress+HauserReparaturset Deckel Getrenntausführung Edelstahl

Werkzeugliste

8 mm

Warnhinweissymbole
GEFAHR
A0011189-DE

WARNUNG
A0011190-DE

VORSICHT
A0011191-DE

HINWEIS
A0011192-DE

GEFAHR!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht vermieden wird, zu Tod oderschwerer Körperverletzung führen wird.
WARNUNG!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht vermieden wird, zu Tod oderschwerer Körperverletzung führen kann.
VORSICHT!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht vermieden wird, zu leichter odermittelschwerer Körperverletzung führen kann.
HINWEIS!
Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden Sachverhalten, die keine
Körperverletzung nach sich ziehen.

Symbole für Informationstypen
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
A0011182

Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
A0011184

A0011193

1.

2.

Endress+Hauser

3.

TippKennzeichnet zusätzliche Informationen.
Handlungsschritte

3Reparaturset Deckel Getrenntausführung Edelstahl

Austausch Deckel
HINWEISDen Deckel öffnen.

2.

Originaldeckel ist demontiert.

3.

Sicher stellen, dass der defekte Deckel inklusive Dichtung entsorgt wird (siehe Bild 3 unten). Keine Wiederverwendung!

4.

Etikett „Use exclusively Cover Revision 1“ im zu reparierenden Gehäuse anbringen (siehe Bild 4 unten).

5.

Neuer Deckel ist bereit für die Montage.

6.

Neuen Deckel schliessen und Schrauben mit mindestens 3.5 Nm anziehen.
3.
21319480-0000A

yelge

br

88 mmmm

gn

2.
whws

1.

whws

5! Cable.= dN.cisolate
beN.c.

7brw
w hsc.

N.

537

20A00-080c.

4N.

shleabC dte. = la.c o
N e isb794

31!
EP

1ils

41

Co

42

gn

0A
0-000

31948

c.

N.

yege lN.c.

37r ernso
Se ehmDufn
EP to ssa
s Meeldshiyel
ge

2
1N.c.

42 41

5 7 4 37!

gn

4N.c.

N.c.

N.c.

brCoils

EPD

Dields42

to Moilsrr e
so hm
en e
S ufnsaes

C

41

N.c. = Cable shields toSensor
be isolated
Messaufnehmer

A0012879

A0027065

A0027205

319480-0000A

2gn

yel
ge

br

5.
whws

4.

1

1.

seU
ely
sivcluexerov
Cnioisev
R
1

N.c.

N.c.

N.c.

5 7 4 37w
w hs

br!

c.
N.

ye
ge l

gnc.
N.

50A00
-080
94
31!
74c.
N.

shleabC dte. = la.c o
N e isb

1

237sldieDPEsoilC1 rn eio or 4m
isns 0e8heve 9
R S 130ufn
7 saes41

A0028643

42to MUse exclusively Cover Revision 1EPD

42 41
Coils

N.c. = Cable shields to Revision 1Sensor
be isolated
71309084
Messaufnehmer
A0028644

6.

88 mm
mm
>3,5 Nm(>2.6 lbf ft)

A0028646

4Endress+Hauser