EA00082D/06/A2/11.19
71430401
2019-04-01

Products

Solutions

Installation Instructions
Replacing the transmitter
Promag 10, 23, 50, 53, 55

EN Installation Instructions: EA00082D →  3
DE Einbauanleitung: EA00082D →  21

Services

EA00082D

Replacing the transmitter
Promag 10, 23, 50, 53, 55
Table of contents
1

Overview of replacement transmitters . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

3Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

5Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

6

Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

7Promag 10 compact version and remote version . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

8Promag 23 compact version . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11

9

Promag 50, 53, 55 compact version and remote version . . . . . . . . . . . . . . . . . . . . . . . . . . . 13

Endress+Hauser

3EA00082D

1

Overview of replacement transmitters

The Installation Instructions apply to the following replacement transmitters:
Order number

Original replacement transmitter

10XXX-

Promag 10 transmitter, aluminum

23XXX-

Promag 23 transmitter, aluminum

50XXX-

Promag 50 transmitter, aluminum

50XXX-

Promag 50 transmitter, stainless steel

50XXX-

Promag 50 transmitter, wall-mount housing

53XXX-

Promag 53 transmitter, aluminum

53XXX-

Promag 53 transmitter, stainless steel

53XXX-

Promag 53 transmitter, wall-mount housing

55XXX-

Promag 55 transmitter, aluminum

55XXX-

Promag 55 transmitter, stainless steel

55XXX-

Promag 55 transmitter, wall-mount housing• The order number of the spare parts set (on the product label on the package) candiffer from the production number (on the label directly on the spare part)!
• You can find the order number of the relevant spare parts set by entering the
production number of the spare part in the spare parts search tool.
• We recommend that you keep the Installation Instructions and packaging together at
all times.

4

Endress+Hauser

EA00082D

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

Order code

4.

=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

3

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

Measuring device approval

Personnel authorized to perform repairs 1)

With approval (e.g. IECEx)

2, 3

Without approval

2, 3

For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Endress+Hauser

5EA00082D

4

Safety instructions• Check whether the spare part matches the labeling on the measuring device as described on
the cover page.
• The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:– Specialized technical staff must be trained in instrument safety.
– They must be familiar with the individual operating conditions of the devices.
– In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is deenergized.
• When using Ex-certified measuring devices: Open device only when in a deenergized state(allow 10 minutes to elapse after switching off the power supply) or in environments which
do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: Commission in accordance with Operating Instructions after
repair. Document the repair procedure.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: Allow the system and measuring
device to cool down to a touchable temperature.
• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removal of the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn instrument off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open housing for a brief period. Avoid the penetration of foreign bodies, moisture orcontaminants.
• Printed circuit boards for devices in categories II1G, II1D, II 2G, II 2D, Zone 1 and Class I, Div.
1, Gp. A, B, C, D have a recess / printed circuit board codification. Unauthorized breaking of
this codification is not permitted as explosion protection will then no longer apply.

6

Endress+Hauser

EA00082D

Example of codification, Ex board

Example of codification, non-Ex board• Caution! When replacing the amplifier board, I/O board or submodules, ensurecompatibility with the installed software. The procedure for reading out the software
version number is described in the Operating Instructions (device functions). If the printed
circuit board's software is not compatible, an update must be performed using an operating
software tool (e.g. Field Care). Notify the plant operator in the event of functional changes.
• Replace defect seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.
• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:– Do not connect in explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

5

Symbols used

5.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
Endress+Hauser

7EA00082D

NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

5.2

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Tools list

3 mm

T20

PH2

Pin/tool

Acid-free, non-Have sealing ringsØ ≤ 2 mm (0.08 inch)

hardening lubricant

available

8

0.5 x 3.5 mm

8 mm, 10 mm

Endress+Hauser

EA00082D

7

Promag 10 compact version and remote version

7.1

Replacing the transmitter

LWARNING
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is de-energized.
Remove the housing seal. When installing, use a new housing seal.
1→

2→

3→

2.
1.

1.

4

0,5 × 3,5 mm

1.
2.

3.

Endress+Hauser

4.

9EA00082D

Special screw! Do not loosen the screw completely or replace it with another one. Only
use original parts from Endress+Hauser.
1→

2

2.
1.

12 Nm

8 mm

‣ Turn neck component and releasescrews.

7.1.1‣ Turn the transmitter and lift it
off.

Reassembling the transmitter housing

Reassembly is carried out in reverse order. Note the following:
When installing, use new housing seals!
1→2‣ Acid-free, non-hardening
lubricantFor detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

10

Endress+Hauser

EA00082D

8

Promag 23 compact version

8.1

Replacing the transmitter

LWARNING
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is de-energized.
Remove the housing seal. When installing, use a new housing seal.
1→

2→

3→

0,5 × 3,5 mm

1.

4.

2.

3.
3.

4.
2.

4→

1.

Ph 2

5

1.
3.

4.

S-DAT

T-DAT

2.
3 mm

Endress+Hauser

11

EA00082D

Special screw! Do not loosen the screw completely or replace it with another one. Only
use original parts from Endress+Hauser.
1

3.
2.
1.

8 mm
12 Nm
(8.9 lbf ft)‣ Release screws, turn thetransmitter and lift it off.

8.1.1

Reassembling the transmitter housing

Reassembly is carried out in reverse order. Note the following:
When installing, use new housing seals!
1→2‣ Acid-free, non-hardening
lubricantFor detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

12

Endress+Hauser

EA00082D

9

Promag 50, 53, 55 compact version and remoteversion

9.1

Removing electronics module from aluminum, stainless steeltransmitter

LWARNING
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is de-energized.
Remove the housing seal. When installing, use a new housing seal.
1→

2→

2.

4.

3.

2.

4.

3.

5.

3.
5.

1.

2.

3 mm

1.

8 mm

‣ Open non-Ex version and Ex version of aluminum transmitter‣ Open stainless steel transmitter

3→

5→

4→

0,5 × 3,5 mm

1.

1.

T20

2.
2.
3.

Endress+Hauser

4.

3.

S-DAT

2.

T-DAT

13

EA00082D

6→

7→

8

1.
T20

Ø < 2 mm2.
‣ In Category II2GD and Div.1

devices for use in hazardous
areas, shielding plates are locatedbetween the individual
electronics boards. Remove theseand insert them again when
installing.

9.2

Replacing non-Ex version of aluminum transmitter

Proceed as described in Section 9.1 →  13and as illustrated in the pictures below.
Special screw! Do not loosen the screw completely or replace it with another one. Only
use original parts from Endress+Hauser.
1→

2

2.

2.

1.
1.

8 mm
7 Nm
(5.2 lbf ft)‣ Turn neck component and releasescrews.

14‣ Turn the transmitter and lift it
off.

Endress+Hauser

EA00082D

9.2.1

Assembling non-Ex version of aluminum transmitter

Reassembly is carried out in reverse order. Note the following:
When installing, use new housing seals!
1→

2→

3

1.

2.

PH 2
0.5 × 3.5 mm‣ Acid-free, non-hardening
lubricantFor detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

9.3

Replacing Ex version of aluminum transmitter

Proceed as described in Section 9.1 →  13and as illustrated in the pictures below.
1→

2→

3

2.
1.

4 mm
7 Nm
(5.2 lbf ft)‣ Release the lock screw on thetransmitter.

Endress+Hauser

1.
2.

3 mm
4 Nm
(2.95 lbf ft)‣ Release the Allen screws on themetal cover of the transmitter,and remove the metal cover.

‣ Turn the transmitter and lift it
off.

15

EA00082D

9.3.1

Assembling Ex version of aluminum transmitter

Reassembly is carried out in reverse order. Note the following:
When installing, use new housing seals!
1→

2→

3→

‣ Acid-free, non-hardening‣ Acid-free, non-hardening

lubricantlubricant

4

1.

2.

PH 2
0.5 × 3.5 mm

For detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

16

Endress+Hauser

EA00082D

9.4

Replacing the stainless steel transmitter

Proceed as described in Section 9.1 →  13and as illustrated in the pictures below.
1→

2

2.
12 Nm
(8.9 lbf ft)

1.

8 mm
10
mm

‣ Release the screws.

9.4.1‣ Turn the transmitter and lift it
off.

Assembling the stainless steel transmitter

Reassembly is carried out in reverse order. Note the following:
When installing, use new housing seals!
1→

2→

3

1.

2.

PH 2
0.5 × 3.5 mm‣ Acid-free, non-hardening
lubricantFor detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

Endress+Hauser

17

EA00082D

9.5

Replacing the transmitter wall-mount housing

LWARNING
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is de-energized.
Remove the housing seal. When installing, use a new housing seal.
1→

2→

3→

0,5 × 3,5 mm

1.

PH2

1.

T20

2.

2.

3.
PH2

4.

3.
4.

4

1.
PH1

2.

18

Endress+Hauser

EA00082D

9.5.1

Assembling the transmitter wall-mount housing

Reassembly is carried out in reverse order. Note the following:
When installing, use new housing seals!
1→

2

1.

2.

PH 2
0.5 × 3.5 mm‣ Acid-free, non-hardening
lubricantFor detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

Endress+Hauser

19

EA00082D

Austausch Messumformer
Promag 10, 23, 50, 53, 55
Inhaltsverzeichnis
1Übersicht Ersatzmessumformer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23

3

Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

5

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25

6

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26

7

Promag 10 Kompaktausführung und Getrenntausführung . . . . . . . . . . . . . . . . . . . . . . . . . 27

8

Promag 23 Kompaktausführung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29

9

Promag 50, 53, 55 Kompaktausführung und Getrenntausführung . . . . . . . . . . . . . . . . . . 31

Endress+Hauser

21

EA00082D

1Übersicht Ersatzmessumformer

Die Einbauanleitung ist für folgende Ersatzmessumformer gültig:
Bestellnummer

Original Ersatzmessumformer

10XXX-

Messumformer Alu Promag 10

23XXX-

Messumformer Alu Promag 23

50XXX-

Messumformer Alu Promag 50

50XXX-

Messumformer Edelstahl Promag 50

50XXX-

Messumformer Wandgehäuse Promag 50

53XXX-

Messumformer Alu Promag 53

53XXX-

Messumformer Edelstahl Promag 53

53XXX-

Messumformer Wandgehäuse Promag 53

55XXX-

Messumformer Alu Promag 55

55XXX-

Messumformer Edelstahl Promag 55

55XXX-

Messumformer Wandgehäuse Promag 55• Die Bestellnummer des Ersatzteilsets (auf dem Produktaufkleber der Verpackung)kann sich von der Produktionsnummer (auf dem Aufkleber direkt auf dem Ersatzteil)unterscheiden!
• Durch Eingabe der Produktionsnummer des Ersatzteiles im Ersatzteilfindetool kanndie Bestellnummer des entsprechenden Ersatzteilsets ermittelt werden.
• Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

22

Endress+Hauser

EA00082D

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

3

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die
Sicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Mit Zulassung (z.B. IECEx)

2, 3

Ohne Zulassung

2, 3

Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Endress+Hauser

23

EA00082D

4

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosem Zustand (nach Berücksichtigung
einer Wartezeit von 10 Minuten nach Abschalten der Energiezufuhr) oder in Umgebungenöffnen, die keine explosionsfähige Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Leiterplatten für Geräte der Kategorien II1G, II1D, II 2G, II 2D, Zone 1 und Class I, Div. 1, Gp.
A, B, C, D besitzen eine Aussparung / Leiterplattenkodierung. Das eigenmächtige
Herausbrechen der Kodierung ist nicht zulässig, weil dadurch der Explosionsschutz
aufgehoben wird.

24

Endress+Hauser

EA00082D

Beispiel Kodierung Ex-Platine

Beispiel Kodierung Nicht-Ex-Platine• Achtung! Beim Auswechseln von Messverstärker-, I/O-Platine oder Submodulen:
Kompatibilität mit der vorhandenen Software sicherstellen. Der Auslesevorgang derSoftware Revisionsnummer ist in der Betriebsanleitung (Gerätefunktionen) beschrieben.
Wenn die Software der Platine nicht kompatibel ist, muss mit Hilfe einer Bediensoftware (z.
B. Field Care) ein Update durchgeführt werden. Bei funktionalen ÄnderungenAnlagenbetreiber informieren.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

5

Verwendete Symbole

5.1

Warnhinweissymbole

GEFAHR

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen wird.
WARNUNG

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen kann.
VORSICHT

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu leichter oder mittelschwerer Körperverletzung führen kann.
Endress+Hauser

25

EA00082D

HINWEIS

Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden
Sachverhalten, die keine Körperverletzung nach sich ziehen.

5.2

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

6

Handlungsschritte

Werkzeugliste

3 mm

T20

PH2

Stift / Werkzeug

säurefreies, nicht

DichtungsringeØ ≤ 2 mm (0,08 inch)

härtendes Fettbereitstellen

26

0.5 x 3.5 mm

8 mm, 10 mm

Endress+Hauser

EA00082D

7

Promag 10 Kompaktausführung undGetrenntausführung

7.1

Austausch Messumformer

LWARNUNG
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
Gehäusedichtung entfernen. Beim Einbau neue Gehäusedichtung verwenden.
1→

2→

3→

2.
1.

1.

4

0,5 × 3,5 mm

1.
2.

3.

Endress+Hauser

4.

27

EA00082D

Spezialschraube! Schraube nicht ganz lösen oder durch eine andere ersetzen. Nur
Originalteile von Endress+Hauser verwenden.
1→

2

2.
1.

12 Nm

8 mm

‣ Halsteil drehen und Schrauben
lösen.

7.1.1‣ Messumformer drehen undhochziehen.

Zusammenbau Messumformergehäuse

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Beim Einbau neue Gehäusedichtungen verwenden!
1→

2‣ Säurefreies, nicht härtendes Fett

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

28

Endress+Hauser

EA00082D

8

Promag 23 Kompaktausführung

8.1

Austausch Messumformer

LWARNUNG
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
Gehäusedichtung entfernen. Beim Einbau neue Gehäusedichtung verwenden.
1→

2→

3→

0,5 × 3,5 mm

1.

4.

2.

3.
3.

4.
2.

4→

1.

Ph 2

5

1.
3.

4.

S-DAT

T-DAT

2.
3 mm

Endress+Hauser

29

EA00082D

Spezialschraube! Schraube nicht ganz lösen oder durch eine andere ersetzen. Nur
Originalteile von Endress+Hauser verwenden.
1

3.
2.
1.

8 mm
12 Nm
(8.9 lbf ft)‣ Schrauben lösen, Messumformer
drehen und hochziehen.

8.1.1

Zusammenbau Messumformergehäuse

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Beim Einbau neue Gehäusedichtungen verwenden!
1→

2‣ Säurefreies, nicht härtendes Fett

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

30

Endress+Hauser

EA00082D

9

Promag 50, 53, 55 Kompaktausführung undGetrenntausführung

9.1

Ausbau Elektronik Messumformer Alu, Edelstahl

LWARNUNG
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
Gehäusedichtung entfernen. Beim Einbau neue Gehäusedichtung verwenden.
1→

2→

2.

4.

3.

2.

4.

3.

5.

3.
5.

1.

2.

3 mm

1.

8 mm

‣ Öffnen Messumformer Alu in nicht explosionsgeschützter Ausführung und‣ Öffnen Messumformer Edelstahl

3→5→

in explosionsgeschützter Ausführung

4→0,5 × 3,5 mm

1.

1.

T20

2.
2.
3.

Endress+Hauser

4.

3.

S-DAT

2.

T-DAT

31

EA00082D

6→

7→

8

1.
T20

Ø < 2 mm2.
‣ Bei Geräten in

explosionsgeschützter
Ausführung der Kategorie II2GD
und Div.1 sind zwischen deneinzelnen Elektronikplatinen
Abschirmbleche. Diese entfernen
und beim Einbau wieder
einsetzen.

9.2

Austausch Messumformer Alu in nicht explosionsgeschützter
Ausführung

Vorgehen wie in Kap. 9.1 →  31 und wie in den Bildern unten.
Spezialschraube! Schraube nicht ganz lösen oder durch eine andere ersetzen. Nur
Originalteile von Endress+Hauser verwenden.
1→

2

2.

2.

1.
1.

8 mm
7 Nm
(5.2 lbf ft)‣ Halsteil drehen und Schrauben
lösen.

32‣ Messumformer drehen undhochziehen.

Endress+Hauser

EA00082D

9.2.1

Zusammenbau Messumformer Alu in nicht explosionsgeschützter Ausführung

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Beim Einbau neue Gehäusedichtungen verwenden!
1→

2→

3

1.

2.

PH 2
0.5 × 3.5 mm‣ Säurefreies, nicht härtendes Fett

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

9.3

Austausch Messumformer Alu in explosionsgeschützter
Ausführung

Vorgehen wie in Kap. 9.1 →  31 und wie in den Bildern unten.
1→

2→

3

2.
1.

4 mm
7 Nm
(5.2 lbf ft)‣ Sicherungsschraube am
Messumformer lösen.

Endress+Hauser

1.
2.

3 mm
4 Nm
(2.95 lbf ft)‣ Imbusschrauben an derMetallabdeckung desMessumformers lösen,
Metallabdeckung abnehmen.

‣ Messumformer drehen undhochziehen.

33

EA00082D

9.3.1

Zusammenbau Messumformer Alu in explosionsgeschützter Ausführung

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Beim Einbau neue Gehäusedichtungen verwenden!
1→

2→

3→

‣ Säurefreies, nicht härtendes Fett

‣ Säurefreies, nicht härtendes Fett

4

1.

2.

PH 2
0.5 × 3.5 mm

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

34

Endress+Hauser

EA00082D

9.4

Austausch Messumformer Edelstahl

Vorgehen wie in Kap. 9.1 →  31 und wie in den Bildern unten.
1→

2

2.
12 Nm
(8.9 lbf ft)

1.

8 mm
10
mm

‣ Schrauben lösen.

9.4.1‣ Messumformer drehen undhochziehen.

Zusammenbau Messumformer Edelstahl

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Beim Einbau neue Gehäusedichtungen verwenden!
1→

2→

3

1.

2.

PH 2
0.5 × 3.5 mm‣ Säurefreies, nicht härtendes Fett

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

Endress+Hauser

35

EA00082D

9.5

Austausch Messumformer Wandgehäuse

LWARNUNG
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
Gehäusedichtung entfernen. Beim Einbau neue Gehäusedichtung verwenden.
1→

2→

3→

0,5 × 3,5 mm

1.

PH2

1.

T20

2.

2.

3.
PH2

4.

3.
4.

4

1.
PH1

2.

36

Endress+Hauser

EA00082D

9.5.1

Zusammenbau Messumformer Wandgehäuse

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Beim Einbau neue Gehäusedichtungen verwenden!
1→

2

1.

2.

PH 2
0.5 × 3.5 mm‣ Säurefreies, nicht härtendes Fett

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

Endress+Hauser37

*71430401*
71430401

www.addresses.endress.com