EA01176G/00/A2/02.18
71424807
2018-12-14*71424807*71424807

Solutions

ProductsServices

Installation Instructions
Cover, cover clamp, O-ringMicropilot NMR8x
Proservo NMS8x
Tankside Monitor NRF81

Overview
This installation instruction is valid for the following spare part sets:
Order number

Original spare part set

Usage

71312596

Front cover Alu, sight glas, gasket

NMR81/84, NRF81, NMS80/81/83

71312607

Front cover 316L, sight glas, gasket

NMR81/84, NRF81, NMS81

71312627

Cover clamp, 2 pieces, TG

NMR81/84, NRF81, NMS80/81/83

71312582

O-ring, housing, 177.39 x 3.53, TG

NMR81/84, NRF81, NMS80/81/83

We recommend that you keep the Installation Instructions and packaging together at all times.

Designated use
The spare part sets and Installation Instructions are used to replace a faulty unit with a functioning unit of the same type. Onlyoriginal parts from Endress+Hauser may be used. Only spare parts kits designed by Endress+Hauser for the measuring device canbe used at any time.
Inspection is performed using the W@M Device Viewer. The procedure involved is described as follows.
A spare parts overview is located in the interior of some measuring devices. If the spare parts kit is specified on this list, noinspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code

=Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for product data and click
Spare Parts.

3.

All spare parts for the measuring device are displayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order number of the The
spare parts kit corresponds to an order number on the spareparts list.

Personnel authorized to carry out repairs
Authorization to carry out repairs depends on the measuring device's approval type. The table below shows the authorized groupof people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out safely and to the required quality
standard. He/she must also guarantee the safety of the device following repair.

EA01176G

Measuring device approval

Authorized repair personnel 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 31)

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser, 3 = Endress+Hauser (return measuringdevice to manufacturer)

Safety instructions• Check whether the spare part matches the labeling on themeasuring device as described on the cover page.
• The spare part sets and Installation Instructions are used toreplace a faulty unit with a functioning unit of the sametype.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting,electrical installation, commissioning, maintenance andrepair procedures.
• The following requirements must be met with regard tospecialized technical staff for the mounting, electricalinstallation, commissioning, maintenance and repair of themeasuring devices:– Specialized technical staff must be trained in instrument
safety.
– They must be familiar with the individual operating
conditions of the devices.
– In the case of Ex-certified measuring devices, they mustalso be trained in explosion protection.
• The measuring device is energized! Risk of fatal injury fromelectric shock. Open the measuring device only when thedevice is de-energized.
• When using Ex-certified measuring devices: Open device
only when in a de-energized state (allow 10 minutes toelapse after switching off the power supply) or inenvironments which do not have a potentially explosive
atmosphere.
• In the case of measuring devices in safety-related
applications in accordance with IEC 61508 or IEC 61511:
Commission in accordance with Operating Instructions afterrepair. Document the repair procedure.
• Before removing the device: set the process to a safe stateand purge the pipe of dangerous process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow
the system and measuring device to cool down to a
touchable temperature.

• In the case of devices in custody transfer, the custodytransfer status no longer applies once the seal has been
removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you
have a working environment protected from electrostatic
discharge.
• After removing the electronics compartment cover: risk ofelectrical shock due to missing touch protection!
Turn the measuring device off before removing internalcovers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid thepenetration of foreign bodies, moisture or contaminants.
• Replace defective seals only with original seals fromEndress+Hauser.
• If threads are damaged or defective, the measuring device
must be repaired.
• Threads (e.g. of the electronics compartment cover andconnection compartment cover) must be lubricated if anabrasion-proof dry lubricant is not available. Use acid-free,
non-hardening lubricant.
• If spacing is reduced or the dielectric strength of themeasuring device cannot be guaranteed during repair work,
perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:– Do not connect in explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning thedevice outlined in the Operating Instructions.
If you have any questions, please contact yourEndress+Hauser service organization.

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
ForbiddenProcedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

2

Series of steps

Endress+Hauser

EA01176G

Tool list

Loosen seal (for custody transfer applications only)
1

Exchange cover
1→

2→

3

Assembly has to be done in reverse order.

Endress+Hauser

3

EA01176G

Exchange cover clamp
1→

2

Assembly has to be done in reverse order.

Exchange O-ring
1→

2→

3→

4

Assembly has to be done in reverse order.

Seal device (for custody transfer applications only)
1

4

Endress+Hauser

EA01176G/00/A2/02.18
71424807
2018-12-14*71424807*
71424807

Solutions

ProductsServices

Einbauanleitung
Deckel, Deckelsicherung, Dichtung
Micropilot NMR8x
Proservo NMS8x
Tankside Monitor NRF81ÜbersichtDie Anleitung ist für folgende Ersatzteilsets gültig:
Bestell-Nummer

Original Ersatzteilset

Verwendung

71312596

Deckel Front Alu, Schauglas, Dichtung

NMR81/84, NRF81, NMS80/81/83

71312607

Deckel Front 316L, Schauglas, Dichtung

NMR81/84, NRF81, NMS81

71312627

Deckelsicherung, 2 Stück, TG

NMR81/84, NRF81, NMS80/81/83

71312582

O-Ring, Gehäuse, 177,39 x 3,53, TG

NMR81/84, NRF81, NMS80/81/83

Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

Bestimmungsgemäße Verwendung
Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen eine funktionierende Einheit des gleichen Typs zuersetzen. Es dürfen nur Originalteile von Endress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht. Ist das Ersatzteilset dort aufgelistet,entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code

=Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten suchen undauf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

wenn die Bestellnummer des Ersatzteilsets mit einer
 NurBestellnummer in der Ersatzteilliste übereinstimmt, darf das
Ersatzteilset verwendet werden.

Reparaturberechtigte Personen
Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts abhängig. Die Tabelle zeigt den jeweils
berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die Sicherheit während der Arbeiten, die
Qualität der Ausführung und die Sicherheit des Geräts nach der Reparatur.

EA01176GZulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3Mit Zulassung (z.B. IECEx)

1, 2, 31)

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker, 3 = Endress+Hauser (Messgerät anHersteller zurücksenden)

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf
dem Messgerät passt, wie auf der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte
Einheit gegen eine funktionierende Einheit des gleichenTyps zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen
Installation, Inbetriebnahme, Wartung und Reparatur
einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage,elektrische Installation, Inbetriebnahme, Wartung undReparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Gerätevertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich imExplosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch
Stromschlag. Messgerät nur im spannungslosen Zustandöffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosemZustand (nach Berücksichtigung einer Wartezeit von 10
Minuten nach Abschalten der Energiezufuhr) oder in
Umgebungen öffnen, die keine explosionsfähige
Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen
gemäß IEC 61508 bzw. IEC 61511: Nach Reparatur
Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustandbringen und Leitung von gefährlichen Prozessstoffen
befreien.
• Verbrennungsgefahr durch heiße Oberflächen! VorArbeitsbeginn: Anlage und Messgerät auf
berührungssichere Temperatur abkühlen.

• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach
Entfernen der Plombe ist der geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESDgeschützte Arbeitsumgebung herstellen.
• Nach Entfernen der Elektronikabdeckung:
Stromschlaggefahr durch aufgehobenen Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt
werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen vonFremdkörpern, Feuchtigkeit oder Verunreinigung
vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen vonEndress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung desMessgeräts.
• Gewinde (z.B. von Elektronikraum- undAnschlussraumdeckel) müssen geschmiert sein, sofern keine
abriebfeste Trockenschmierung vorhanden ist. Säurefreies,
nicht härtendes Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder
die Spannungsfestigkeit des Messgeräts nicht sichergestellt
werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß
Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum
Transport und zur Rücksendung beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige
Endress+Hauser Serviceorganisation.

Symbole für Informationstypen
SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

2

Handlungsschritte

Endress+Hauser

EA01176G

Werkzeugliste

Verplombung lösen (nur für Anwendungen im eichpflichtigen Verkehr)
1

Deckel ersetzen
1→

2→

3

Der Einbau erfolgt in umgekehrter Reihenfolge.

Endress+Hauser

3

EA01176G

Deckelsicherung ersetzen
1→2

Der Einbau erfolgt in umgekehrter Reihenfolge.

Dichtung ersetzen
1→2→

3→

4

Der Einbau erfolgt in umgekehrter Reihenfolge.

Gerät verplomben (nur für Anwendungen im eichpflichtigen Verkehr)
1

4Endress+Hauser