Torque Gauge

Digital Torque Gauge

ATGE-G (Global model)
PAT. PEND

3 Way Configuration:
Hand-held, Table Top, or as a Torque Meter with Testing Fixture

Hand-held digital torquegauge with self containeddisplay

Pull out display allows fortable top configuration
with measurement stand(Sold separately)

Use as a torque meter withaddition of testing fixture(Sold separately)

Digital torque gauge with pull out display
Ideal for torque measurement of electronic components
Can be used for checking accuracy of torque screwdrivers
4 models cover torque capacities from 0.1cN·m to 5cN·m
Accuracy ±2% +1 digit of indicated value, CE & RoHS compliant
999 data memory storage with USB output
Multiple units of measure capabilities: SI, Metric, AmericanDirection

Digital Torque Gauge

ATGE-GFeatures

3 Way Configuration: Hand-Held, Table Top, or as a Torque Meter with Testing Fixture
Pull out display allows for a variety of testing
configurations: hand held, table top or with testing
fixture. The digital display simply slides out of the
housing and is attached with a curly cord. Use theoptional testing stand to secure the gauge while
testing or add fixture to secure your test piece.
Ideal for torque measurement for electronic
components such as small motor, encoder,potentiometer and others.
4 models cover torque capacities from 0.1cN·m(1mN·m) to 5cN·m (50mN·m).

3 Way Configuration

1.
Hand-held stylewith self contained display* Example: Torque measurement
for small motor, development of
the motor, inspection for control
software2.

Accuracy ±2% +1 digit of indicated value, Compliant
with CE and RoHS directive.
Additional power charging option when using ACadapter (Model: BA-5) and USB cable (Model: No. 384).
The stored data can betransferred to PC with "ExcelReceiver" software (Optional).

Table top stylewith display on the side
* Example: Measurement forstarting torque and click torquefor encoder

3.
Torque Meter stylewith testing fixture* Example: Rotating torquemeasurement for potentiometer

Specifications

ATGE-G Models
Model

Accuracy ±2%+1digitcN·mMin.-Max.
1 digitmN·m
Min.-Max.
1 digitgf·cm
Min.-Max.
1 digitozf·inMin.-Max.
1 digit

ATGE05CN-G
ATGE1CN-G

0.1-0.5
0.2-1

0.001
0.001

1-52-100.01
0.01

10-50
20-1000.1
0.1

0.15-0.7
0.3-1.4

0.001
0.001

ATGE2CN-G
ATGE5CN-G

0.4-2
1-5

0.002
0.005

4-20
10-50

0.02
0.05

40-200100-500

0.2
0.5

0.6-2.8
1.5-7

0.002
0.005

ATGE10CN-G
ATGE20CN-G

2-10
4-20

0.01
0.02

20-10040-2000.1
0.2

200-1000
400-2000

123-14
6-28

0.01
0.02

Optional Accessories
Testing Fixture (No. 800)
* Clump diameter:
ø10 to ø58mm

Measurement
Stand (No. 808)

[Example]

Standard Accessories: Coin-type lithium battery (built-in the body), Carrying case

Dimensions
Chuck Dia.
[mm]
ø1-6.5

Dimensions [mm]Overall Length
Outer Dia.
12067

Weight[kg]0.305

Common Specifications
CW/CCW±2% of indicated value +1 digit
7 segment LCD display, Torque unit, Battery indicator, Direction, Counter 3 digits(character height 3mm), Torque value: 4 digits (character height 7mm)
Mode
PEAK/RUN
Data Memory
999 readings
Statistic Prosessing
Sample size, Max. value, Min. value, Mean value
Data Output
USB output (USB mini B connector)
Power
Coin-type lithium battery (CR2450)
Continuous in Use
Approx. 10 hours when using coin battery
Other Functions
Auto power off (3 min.), Auto memory reset (0.5-5) seconds variable, Auto zeroadjustment, Residual battery indicator (4 steps), Buzzer ON/OFF, Unit conversion
Operating Temperature 0 to 40°C non condensingStandard Options
Coin battery (built in), Carrying case

Calibration Kit
ATGTCL24CN

Connecting Cable (No. 384)
* No. 384 cable is required
when using BA-5.

AC Adapter
BA-5

Direction
Accuracy
Display

Calibration Adapter(No. 806)

Resin Chuck (No. 322)
* Scratch-prevention
for an inspection
object.

11.10D