EA01185G/00/A2/01.16
71335165*71335165*71335165

ProductsSolutions

Services

Installation Instructions
Display set, display holder
Micropilot NMR8x
Proservo NMS8x
Tankside Monitor NRF81

Overview
This installation instruction is valid for the following spare part sets:
Order number

Original spare part set

Usage

71312638

Display set, RS485, holder, TG

NMR81/84, NRF81, NMS80/81/83

71312640

Display holder, fixing ring, TG

NMR81/84, NRF81, NMS80/81/83

We recommend that the Installation Instructions are kept with the packaging at all times.

Designated use
The spare part set and the Installation Instruction are used to replace a faulty unit with a functioning unit of the same type. Use
genuine parts from Endress+Hauser only.
Only original spare part sets, intended by Endress+Hauser for the measuring device, must be used.
The verification has to be done via W@M Device Viewer; this procedure is explained below.
For some devices there is an overview of spare part sets inside the device. If the spare part set is listed there, the verification
is not required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (Ser. No.), search for product data and clickon "spare parts".

3.

List of all spare parts sets for the device.

4.

Determine the order number of the spare part set.

spare part set can only be used if the order number of the The
spare part set corresponds to the order number on the spare partlist.

EA01185G

Authorized personnel
Authorization to carry out a repair depends on the approval of the measuring device. The table shows the respective group ofpersons for each.
The person who carries out the repair is responsible for safety during the work, the quality of work completed and safety ofthe device after repair.
Approval of the measuring device

Group of persons authorized to carry out repairs 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 31)

1 = Trained customer technician, 2 = Service technician authorized byEndress+Hauser, 3 = Endress+Hauser (send measuring device back tomanufacturer)

Safety instructions• Check whether the spare part matches the identification
label on the measuring device, as explained on the first page.
• The spare parts set and Installation Instructions are used toreplace a faulty unit with a functioning unit of the sametype.
Use genuine parts from Endress+Hauser only.
• Comply with national regulations governing mounting,electrical installation, commissioning, maintenance andrepair procedures.
• Requirements with regard to specialized technical staff forthe mounting, electrical installation, commissioning,maintenance and repair of the measuring devices:– trained in instrument safety.
– familiar with the individual operation conditions of thedevices.
– for Ex-certified measuring devices: also trained inexplosion protection.
• The measuring device is energized. Danger: Risk of electric
shock! Open the measuring device in a de-energized state
only.
• In the case of Ex-certified measuring devices: Only open in ade-energized state (once a delay of 10 minutes has elapsed
after switching off the power supply) or in environments
which do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related
applications in accordance with IEC 61508 or IEC 61511:
After repair recommission in accordance with OperatingInstructions. Document the repair procedure.
• Before removing the device: set the process in a safe
condition and purge the pipe of dangerous materials.
• Hot surfaces! Risk of injury! Before commencing work, allow
the system and measuring device to cool down to a
touchable temperature.

• In the case of measuring devices in custody transfer, thecustody transfer status no longer applies once the lead seal
has been removed.
• Follow the Operating Instructions for the device.
• Risk of damaging electronic components! Ensure you have aworking environment protected from electrostatic discharge.
• After removing the electronics cover, there is a risk ofelectric shock as shock protection is removed!
Switch off the measuring device before removing internalcovers.
• Modifications to the measuring device are not permitted.
• Only open housing for a brief period. Avoid the penetrationof foreign bodies, moisture or contaminants.
• Replace defective seal/gaskets with genuine parts fromEndress+Hauser only.
• If threads are damaged or defective, the measuring device
must be repaired.
• Threads (e.g. of the cover for the electronics and connectioncompartments) must be lubricated. Use an acid-free, nonhardening grease if an abrasion resistant dry lubricant isnon-existent.
• If spacing is reduced or the dielectric strength of themeasuring device cannot be guaranteed during repair work,
perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service connector:– Do not connect in potentially explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning thedevice outlined in the Operating Instructions.
If you have any questions, contact yourEndress+Hauser service organization.

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

2

Endress+Hauser

EA01185G

SymbolMeaning
Reference to documentation

1. , 2. , 3. …

Series of steps

Tool list

Loosen seal (for custody transfer applications only)
1

Remove display set
1→

2→

5→

6

3→

4→

The spare part list for this device is printed on the rear side of the orginal cover. For using the spare part list furthermore, werecommend to use the orginal cover with the printed spare part list again, see chapter "Remove display holder".

Endress+Hauser

3

EA01185G

Remove display holder
7→

8→

81
NMS
ureervotruct
Pros AF4
e no./sfor:3451
Spar
C098
7102 3451
e partsA21A
Spar no.:87102 3451 -AABICR+
7102 0002 EFLALC76 G+
Ser.XPF AEB -AABEFE G+
AAE 0002 -AABEFE G+
XPF 0002 -AABEFE G+
XPF 0002 -AABEFE G+
XPF 0002 -AABEFE G+G+
XPF 0002 -AABEFEXPF 0002 -AABEFE G+
XPF 0002 -AABEFE
XPF 0002
G+
XPF 3689
7127 xxxx -AABEFE+
ADJ
7122 0002 DEFEG+
XPF AAC -AAACAB KP
AAA 0002 HRIJJAJBXPF AHC
AAH

1112,3 4-8 9 10

2,311

Spare parts for: Proservo NMS81
Ser.-no.:8A21AC098AF4
Spare part
1 Cover
2 Display
3 Display asm.
4 Main electr.

Spare parts for: Proservo NMS81
Ser.-no.:8A21AC098AF4
Spare part
1 Cover
2 Display
3 Display asm.
4 Main electr.

Spare no./structure71023451
71023451
71023451
XPF0002-AABICR+AAEAEBEFLALC76
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
71273689
7122xxxx
XPF0002-AABEFEG+AAAAACDEFEG+
12 Wire Drum
XPF0002-AAACABADJ+AAHAHCHRIJJAJBKP

5 IOM-V1
Slot A
6 IOM-A/RTD
Slot B
7 IOM-D
Slot C
8 IO Mod FF
Slot D
9 SMS electr.
10 Detector
11 Displacer

121
2,3 4-8 9 10r:
ationviewe
informevice
ional om/dAddit ess.c.endr
www12ation: iewer
informevicev
nal.com/der

ation:
eview
inform/deviconals.com
Additindres

10Additiondress4-8

Spare no./structure71023451
71023451
71023451
XPF0002-AABICR+AAEAEBEFLALC76
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
71273689
7122xxxx
XPF0002-AABEFEG+AAAAACDEFEG+
12 Wire Drum
XPF0002-AAACABADJ+AAHAHCHRIJJAJBKP

5 IOM-V1
Slot A
6 IOM-A/RTD
Slot B
7 IOM-D
Slot C
8 IO Mod FF
Slot D9 SMS electr.
10 Detector
11 Displacer

www.e9110

Spare parts for: Proservo NMS81
Ser.-no.:8A21AC098AF4
Spare part
1 Cover
2 Display
3 Display asm.
4 Main electr.

12www.er
on: iewemati icev
infor /dev
alition .com
Add ress.end
wwwe partrSpar.
Cove
1 Display asm
2 Displayelectr.
3 Main4-V1IOM
5 Slot A TD
-A/RIOM
6 Slot B-D
IOM
7 Slot C FF
IO Mod
8 Slot D electr.
SMS ctor
9 Dete lacer
10 Disp11DrumWire
129→

Spare no./structure71023451
71023451
71023451
XPF0002-AABICR+AAEAEBEFLALC76
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
71273689
7122xxxx
XPF0002-AABEFEG+AAAAACDEFEG+
12 Wire Drum
XPF0002-AAACABADJ+AAHAHCHRIJJAJBKP

5 IOM-V1
Slot A
6 IOM-A/RTD
Slot B
7 IOM-D
Slot C
8 IO Mod FF
Slot D
9 SMS electr.
10 Detector
11 Displacer

1212,3 4-8 9 101111

The spare part list for this device is printed on the rear side of the original cover. For using the spare part list furthermore,
we recommend to use the orginal cover with the printed spare part list again.

Assembly spare part sets
Assembly has to be done in reverse order.
When inserting the display into the display holder, please observe the following:

OPEN

open
open

1

open
open

OPEN

open

Seal device (for custody transfer applications only)
1

4

Endress+Hauser

EA01185G/00/A2/01.16
71335165*71335165*71335165

ProductsSolutions

Services

Einbauanleitung
Display Set, Displayhalter
Micropilot NMR8x
Proservo NMS8xTankside Monitor NRF81ÜbersichtDie Anleitung ist für folgende Ersatzteilsets gültig:
Bestell-Nummer

Original Ersatzteilset

Verwendung

71312638

Display Set, RS485, Halter, TG

NMR81/84, NRF81, NMS80/81/83

71312640

Displayhalter, Fixierring, TG Plattform

NMR81/84, NRF81, NMS80/81/83

Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

Bestimmungsgemäße Verwendung
Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen eine funktionierende Einheit des gleichen Typs zuersetzen. Es dürfen nur Originalteile von Endress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht. Ist das Ersatzteilset dort aufgelistet,entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten suchen undauf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

wenn die Bestellnummer des Ersatzteilsets mit einer
 NurBestellnummer in der Ersatzteilliste übereinstimmt, darf das
Ersatzteilset verwendet werden.

EA01185G

Reparaturberechtigte Personen
Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts abhängig. Die Tabelle zeigt den jeweilsberechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die Sicherheit während der Arbeiten, dieQualität der Ausführung und die Sicherheit des Geräts nach der Reparatur.

1)

Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3Mit Zulassung (z.B. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker, 3 = Endress+Hauser (Messgerät anHersteller zurücksenden)

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf
dem Messgerät passt, wie auf der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte
Einheit gegen eine funktionierende Einheit des gleichen
Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen
Installation, Inbetriebnahme, Wartung und Reparatur
einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage,elektrische Installation, Inbetriebnahme, Wartung undReparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Gerätevertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im
Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch
Stromschlag. Messgerät nur im spannungslosen Zustandöffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosem
Zustand (nach Berücksichtigung einer Wartezeit von 10
Minuten nach Abschalten der Energiezufuhr) oder in
Umgebungen öffnen, die keine explosionsfähige
Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen
gemäß IEC 61508 bzw. IEC 61511: Nach Reparatur
Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustandbringen und Leitung von gefährlichen Prozessstoffen
befreien.
• Verbrennungsgefahr durch heiße Oberflächen! VorArbeitsbeginn: Anlage und Messgerät auf
berührungssichere Temperatur abkühlen.

• Bei Messgeräten im abrechnungspflichtigen Verkehr: NachEntfernen der Plombe ist der geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESDgeschützte Arbeitsumgebung herstellen.
• Nach Entfernen der Elektronikabdeckung:
Stromschlaggefahr durch aufgehobenen Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entferntwerden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen vonFremdkörpern, Feuchtigkeit oder Verunreinigung
vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen vonEndress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung desMessgeräts.
• Gewinde (z.B. von Elektronikraum- undAnschlussraumdeckel) müssen geschmiert sein, sofern keine
abriebfeste Trockenschmierung vorhanden ist. Säurefreies,
nicht härtendes Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder
die Spannungsfestigkeit des Messgeräts nicht sichergestellt
werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß
Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum
Transport und zur Rücksendung beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige
Endress+Hauser Serviceorganisation.

Symbole für Informationstypen
SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.

2Endress+Hauser

EA01185G

SymbolBedeutung
TippKennzeichnet zusätzliche Informationen.
Verweis auf Dokumentation

1. , 2. , 3. …

Handlungsschritte

Werkzeugliste

Verplombung lösen (nur für Anwendungen im eichpflichtigen Verkehr)
1

Display Set ausbauen
1→

2→

5→

6

3→

4→

Auf der Rückseite der original Abdeckung befindet sich die zum Gerät passende Ersatzteilübersicht. Um dieseErsatzteilübersicht weiterhin nutzen zu können, empfehlen wir weiterhin die originale Abdeckung zu verwenden, sieheKapitel "Austausch des Displayhalters".

Endress+Hauser

3

EA01185GDisplayhalter ausbauen
7→8→

81
NMS
ureervotruct
Pros AF4
e no./sfor:
3451
Spar
C098
7102 3451
e partsA21A
Spar no.:87102 3451 -AABICR+
7102 0002 EFLALC76 G+
Ser.XPF AEB -AABEFE G+
AAE 0002 -AABEFE G+
XPF 0002 -AABEFE G+
XPF 0002 -AABEFE G+
XPF 0002 -AABEFE G+G+
XPF 0002 -AABEFE
XPF 0002 -AABEFE G+
XPF 0002 -AABEFE
XPF 0002
G+
XPF 3689
7127 xxxx -AABEFE+
ADJ
7122 0002 DEFEG+
XPF AAC -AAACAB KP
AAA 0002 HRIJJAJBXPF AHC
AAH

1112,3 4-8 9 102,311

Spare parts for: Proservo NMS81
Ser.-no.:8A21AC098AF4
Spare part
1 Cover
2 Display
3 Display asm.
4 Main electr.

Spare parts for: Proservo NMS81
Ser.-no.:8A21AC098AF4
Spare part
1 Cover
2 Display
3 Display asm.
4 Main electr.

Spare no./structure71023451
71023451
71023451
XPF0002-AABICR+AAEAEBEFLALC76
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
71273689
7122xxxx
XPF0002-AABEFEG+AAAAACDEFEG+
12 Wire Drum
XPF0002-AAACABADJ+AAHAHCHRIJJAJBKP

5 IOM-V1
Slot A
6 IOM-A/RTD
Slot B
7 IOM-D
Slot C
8 IO Mod FF
Slot D
9 SMS electr.
10 Detector
11 Displacer

121
2,3 4-8 9 10r:
ationviewe
informevice
ional om/dAddit ess.c.endr
www12ation: iewer
informevicev
nal.com/der

ation:
eview
inform/devic

onal
s.com
Additindres

10Additiondress4-8

Spare no./structure71023451
71023451
71023451
XPF0002-AABICR+AAEAEBEFLALC76
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
71273689
7122xxxx
XPF0002-AABEFEG+AAAAACDEFEG+
12 Wire Drum
XPF0002-AAACABADJ+AAHAHCHRIJJAJBKP

5 IOM-V1
Slot A
6 IOM-A/RTD
Slot B
7 IOM-D
Slot C
8 IO Mod FF
Slot D
9 SMS electr.
10 Detector
11 Displacer

www.e9110

Spare parts for: Proservo NMS81
Ser.-no.:8A21AC098AF4
Spare part
1 Cover
2 Display
3 Display asm.
4 Main electr.

12www.er
on: iewemati icev
infor /dev
alition .com
Add ress.end
wwwe partr
Spar.
Cove
1 Display asm
2 Displayelectr.
3 Main4-V1IOM
5 Slot A TD
-A/RIOM
6 Slot B-DIOM
7 Slot C FF
IO Mod
8 Slot D electr.
SMS ctor
9 Dete lacer
10 Disp
11DrumWire
129→

Spare no./structure71023451
71023451
71023451
XPF0002-AABICR+AAEAEBEFLALC76
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
XPF0002-AABEFEG+
71273689
7122xxxx
XPF0002-AABEFEG+AAAAACDEFEG+
12 Wire Drum
XPF0002-AAACABADJ+AAHAHCHRIJJAJBKP

5 IOM-V1
Slot A
6 IOM-A/RTD
Slot B
7 IOM-D
Slot C
8 IO Mod FF
Slot D
9 SMS electr.
10 Detector
11 Displacer

1212,3 4-8 9 101111

Auf der Rückseite der original Abdeckung befindet sich die zum Gerät passende Ersatzteilübersicht. Um dieseErsatzteilübersicht weiterhin nutzen zu können, empfehlen wir weiterhin die originale Abdeckung zu verwenden.

Ersatzteile einbauen
Der Einbau erfolgt jeweils in umgekehrter Reihenfolge.
Beim Einbau des Displays in den Halter bitte folgendes beachten:

OPEN

open
open

1

open
open

OPEN

open

Gerät verplomben (nur für Anwendungen im eichpflichtigen Verkehr)
1

4Endress+Hauser