EA01177G/00/A2/02.18
71409170
2018-06-28*71409170*
71409170

Solutions

Products

Services

Installation Instructions
Electronic box complete
Micropilot NMR8x
Proservo NMS8x
Tankside Monitor NRF81

Overview
This installation instruction is valid for the following spare part sets:order number

Original spare part set

Usage

XPN0034-

Electronic box complete NMR8x, NRF81, NMS8x

NMR81/84, NRF81, NMS80/81/83

We recommend that the Installation Instructions are kept with the packaging at all times.

Designated use
The spare part set and the Installation Instruction are used to replace a faulty unit with a functioning unit of the same type. Use
genuine parts from Endress+Hauser only.
Only original spare part sets, intended by Endress+Hauser for the measuring device, must be used.
The verification has to be done via W@M Device Viewer; this procedure is explained below.
For some devices there is an overview of spare part sets inside the device. If the spare part set is listed there, the verification
is not required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=Description?

1.

www.endress.com/deviceviewer2.

Enter serial number (Ser. No.), search for product data and clickon "spare parts".

3.

List of all spare parts sets for the device.

4.

Determine the order number of the spare part set.

spare part set can only be used if the order number of the The
spare part set corresponds to the order number on the spare partlist.

Authorized personnel
Authorization to carry out a repair depends on the approval of the measuring device. The table shows the respective group ofpersons for each.
The person who carries out the repair is responsible for safety during the work, the quality of work completed and safety ofthe device after repair.

EA01177G

Approval of the measuring device

Group of persons authorized to carry out repairs 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 31)

1 = Trained customer technician, 2 = Service technician authorized by Endress+Hauser, 3 = Endress+Hauser (send measuring device backto manufacturer)

Safety instructions• Check whether the spare part matches the identification
label on the measuring device, as explained on the first page.
• The spare parts set and Installation Instructions are used toreplace a faulty unit with a functioning unit of the sametype.
Use genuine parts from Endress+Hauser only.
• Comply with national regulations governing mounting,electrical installation, commissioning, maintenance andrepair procedures.
• Requirements with regard to specialized technical staff forthe mounting, electrical installation, commissioning,maintenance and repair of the measuring devices:– trained in instrument safety.
– familiar with the individual operation conditions of thedevices.
– for Ex-certified measuring devices: also trained inexplosion protection.
• The measuring device is energized. Danger: Risk of electric
shock! Open the measuring device in a de-energized state
only.
• In the case of Ex-certified measuring devices: Only open in ade-energized state (once a delay of 10 minutes has elapsed
after switching off the power supply) or in environments
which do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related
applications in accordance with IEC 61508 or IEC 61511:
After repair recommission in accordance with Operating
Instructions. Document the repair procedure.
• Before removing the device: set the process in a safe
condition and purge the pipe of dangerous materials.
• Hot surfaces! Risk of injury! Before commencing work, allow
the system and measuring device to cool down to a
touchable temperature.

• In the case of measuring devices in custody transfer, thecustody transfer status no longer applies once the lead seal
has been removed.
• Follow the Operating Instructions for the device.
• Risk of damaging electronic components! Ensure you have aworking environment protected from electrostatic discharge.
• After removing the electronics cover, there is a risk ofelectric shock as shock protection is removed!
Switch off the measuring device before removing internalcovers.
• Modifications to the measuring device are not permitted.
• Only open housing for a brief period. Avoid the penetrationof foreign bodies, moisture or contaminants.
• Replace defective seal/gaskets with genuine parts fromEndress+Hauser only.
• If threads are damaged or defective, the measuring device
must be repaired.
• Threads (e.g. of the cover for the electronics and connection
compartments) must be lubricated. Use an acid-free, nonhardening grease if an abrasion resistant dry lubricant isnon-existent.
• If spacing is reduced or the dielectric strength of themeasuring device cannot be guaranteed during repair work,
perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service connector:– Do not connect in potentially explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning thedevice outlined in the Operating Instructions.
If you have any questions, contact yourEndress+Hauser service organization.

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

2

Series of steps

Endress+Hauser

EA01177G

Tool list

Loosen seal (for custody transfer applications only)
1

Remove electronic box
1→

2→

3→

5→

6→

7→

4→

8

Endress+Hauser

3

EA01177G
NMR8x, NMS8x: Take out electronic box
9→10

1. 3x

2.

2.
3.

1.

NRF81: Take out electronic box109→

3xAssembly spare part set
Assembly has to be done in reverse order.

Seal device (for custody transfer applications only)
1→

2

HRon
WP
SIM

4

Endress+Hauser

EA01177G/00/A2/02.18
71409170
2018-06-28*71409170*
71409170

Solutions

Products

Services

Einbauanleitung
Elektronikbox komplett
Micropilot NMR8x
Proservo NMS8x
Tankside Monitor NRF81Übersicht
Die Anleitung ist für folgende Ersatzteilsets gültig:
Bestell-Nummer

Original Ersatzteilset

Verwendung

XPN0034-

Elektronikbox komplett NMR8x, NRF81, NMS8x

NMR81/84, NRF81, NMS80/81/83

Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

Bestimmungsgemäße Verwendung
Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen eine funktionierende Einheit des gleichen Typs zu
ersetzen. Es dürfen nur Originalteile von Endress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht. Ist das Ersatzteilset dort aufgelistet,entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code

=Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten suchen undauf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

wenn die Bestellnummer des Ersatzteilsets mit einer Nur
Bestellnummer in der Ersatzteilliste übereinstimmt, darf dasErsatzteilset verwendet werden.

Reparaturberechtigte Personen
Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts abhängig. Die Tabelle zeigt den jeweils
berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die Sicherheit während der Arbeiten, dieQualität der Ausführung und die Sicherheit des Geräts nach der Reparatur.

EA01177GZulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3Mit Zulassung (z.B. IECEx)

1, 2, 31)

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker, 3 = Endress+Hauser (Messgerät anHersteller zurücksenden)

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf
dem Messgerät passt, wie auf der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte
Einheit gegen eine funktionierende Einheit des gleichen
Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen
Installation, Inbetriebnahme, Wartung und Reparatur
einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage,elektrische Installation, Inbetriebnahme, Wartung undReparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Gerätevertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im
Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch
Stromschlag. Messgerät nur im spannungslosen Zustandöffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosemZustand (nach Berücksichtigung einer Wartezeit von 10Minuten nach Abschalten der Energiezufuhr) oder inUmgebungen öffnen, die keine explosionsfähige
Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen
gemäß IEC 61508 bzw. IEC 61511: Nach Reparatur
Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustandbringen und Leitung von gefährlichen Prozessstoffen
befreien.
• Verbrennungsgefahr durch heiße Oberflächen! VorArbeitsbeginn: Anlage und Messgerät auf
berührungssichere Temperatur abkühlen.

• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach
Entfernen der Plombe ist der geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESDgeschützte Arbeitsumgebung herstellen.
• Nach Entfernen der Elektronikabdeckung:
Stromschlaggefahr durch aufgehobenen Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt
werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen vonFremdkörpern, Feuchtigkeit oder Verunreinigung
vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen vonEndress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung desMessgeräts.
• Gewinde (z.B. von Elektronikraum- undAnschlussraumdeckel) müssen geschmiert sein, sofern keine
abriebfeste Trockenschmierung vorhanden ist. Säurefreies,
nicht härtendes Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder
die Spannungsfestigkeit des Messgeräts nicht sichergestellt
werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß
Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum
Transport und zur Rücksendung beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige
Endress+Hauser Serviceorganisation.

Symbole für Informationstypen
SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

2

Handlungsschritte

Endress+Hauser

EA01177G

Werkzeugliste

Verplombung lösen (nur für Anwendungen im eichpflichtigen Verkehr)
1

Elektronikbox ausbauen
1→

2→

3→

5→6→7→

4→

8

Endress+Hauser

3

EA01177G
NMR8x, NMS8x: Elektronikbox entnehmen
9→10

1. 3x

2.

2.
3.

1.

NRF81: Elektronikbox entnehmen
10

9→

3x

Ersatzteil einbauen
Der Einbau erfolgt in umgekehrter Reihenfolge.

Gerät verplomben (nur für Anwendungen im eichpflichtigen Verkehr)
1→2

HR
onWP
SIM

4

Endress+Hauser