③整理_RunLog_2105Kr_220204.xlsx1_RunLog2105Kr
Run Summary Sheet : CAMAC & LabView
AttFnamestart
stop Header
21/05/12 実験前
bef1 10:41:12 11:01:04
実験中：Beam調整時
21/05/13
ZnSスポット調整Cap
1 6:15:00
R01_ZnS_Zsld
2
R02_Wob0-36-0
3
R03_Au50umR#6
4
R04_Gaf02
5
R05_Gaf03
Gaf:: EBT3 □85x85mm @ Ysld
Gaf焼き

Ender / LogNote
Elapsepg.
コメント
LogNote: E5exp(7)_p.58
0:19:52 SSD:Kr Gain

12
3

6:45:00

mpg
mpg
mpg
mpg
mpg

7:32:55
7:34:01
7:35:08
7:36:30
7:38:09

Wob AuF Kap IC1 PL1 ED

PL2IC2 Air1 Air2SSD

0 3 0 0 0××

2 3 0 0 0 PorS(0) R36 50u 〇〇100u 0u

2 3 0 0 0 PL〇〇〇〇100u var500u

0 0 0 0 0〇〇〇〇〇var

IC2大@Air2PL2@IC2下流

2 3 0 0 0 PorS(0) 〇〇〇〇

〇0u

0 3 0 0 1 ～1 3 0 0 1〇〇

〇

〇

〇0u

IC2大@Air2

0:13:00
0:00:29 1/f=1/999～1/9 手動で変更
0:00:30
0:00:29
0:00:30
0:00:32
0:00:32
0:00:29
0:00:32
0:00:32
0:00:31
0:00:31
0:00:29
0:00:30
10uSv/h0:20:00

0 2 0 0 1 ～1 4 0 0 3〇〇〇〇〇0u

IC2大@Air2

1 4 0 0 3 PorS(0) 〇〇〇〇〇0u1 4 0 0 0 PandS〇〇〇〇var

R=20,36,0145 200p.66
IC2[A]
15s
15s
13s IC1,PL1無し
PL1=100u(new) PL2=500u(前のママ)
Thr PL1=150mV PL2=200mV
PL1=PL2= 1kcps

p.66

PL HV： 利用「前」12345Trig50uΦ50

1:01:00
plhv01-

20 2 1 1 00 2 0 1 00 2 0 0 0

Au 50u, R=0
Au 50u, R=36
Au 50u, R=36 ED出口コリ

回路調整

3p.66Att vs IC2@Air2
Gaf

100 10 5

7:33:32 plhv01-01 500V
7:34:32 plhv01-02 600V
7:36:04 plhv01-03 700V
7:37:40 plhv01-04 800V
7:39:11 plhv01-05 900V

0u

IC2@Xclmt

0:00:37
0:00:31
0:00:56 PL1=700V決定
0:01:10
0:01:02scnEDｐｌ： PL2/PL1 透過率
scnEDpl01_202105130745
7:45:00
scnEDpl011 7:45:16
2 7:45:55
3 7:46:37
4 7:47:19
5 7:48:01
6 7:48:43
7 7:49:25
8 7:50:07
9 7:50:49
10 7:51:31
11 7:52:13
12 7:52:55
13 7:53:37
14 7:54:19
15 7:55:01
16 7:55:43
17 7:56:25
18 7:57:07
19 7:57:49
20 7:58:31
21 7:59:13
22 7:59:55

p.67

PL 1Kcps

8:00:00 30s/点

0:15:00

7:45:43 scanED001 0.0 um
7:46:25 scanED002 100.2 um
7:47:07 scanED003 196.4 um
7:47:49 scanED004 302.1 um
7:48:31 scanED005 397.4 um
7:49:13 scanED006 498.8 um
7:49:55 scanED007 609.2 um
7:50:37 scanED008 735.6 um
7:51:19 scanED009 806.4 um
7:52:01 scanED010 855.0 um
7:52:43 scanED011 859.6 um
7:53:25 scanED012 870.6 um
7:54:07 scanED013 883.4 um
7:54:49 scanED014 893.6 um
7:55:31 scanED015 901.7 um
7:56:13 scanED016 911.9 um
7:56:55 scanED017 922.9 um
7:57:37 scanED018 937.5 um
7:58:19 scanED019 947.6 um
7:59:01 scanED020 955.8 um
7:59:43 scanED021 966.0 um
8:00:25 scanED022 978.8 um

0:00:27
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30 850umくらいから下がる
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30

scnEDic： ExpR 確認
scnEDic01_202105130834
8:34:00

30s/点

PL HV 重照射「後」plhv02-12345

9:15:01
9:16:26
9:17:37
9:19:01
9:20:41p.67

9:09:00

IC1=6.4-8 IC2大=7.4-8A
0:35:00

p.70

9:15:43 plhv02-01 500V aft IC2Bg
9:17:09 plhv02-02 600V PL1nomi
9:18:31 plhv02-03 700V
9:20:13 plhv02-04 800V
9:21:41 plhv02-05 900V

PL1=700cps
0:00:42
0:00:43 IC2大にしたので、殆ど劣化無し
0:00:54
0:01:12
0:01:00

scnAtt： IC1 vs PL1
scnAtt01_202105130944
9:44:00 9:57:00 Step = 1/2, /4, /6, /8, /10
scnAtt011 9:45:53 9:46:22 scanAtt001 Mul 2000
2 9:47:32 9:48:02 scanAtt002 Mul 1000
3 9:48:28 9:48:57 scanAtt003 Mul 2000
4 9:49:28 9:49:58 scanAtt004 Mul 4000
5 9:50:28 9:51:00 scanAtt005 Mul 8000
6 9:51:28 9:52:00 scanAtt006 Mul 10000
7 9:52:28 9:52:57 scanAtt007 Mul 20000
8 9:53:16 9:53:48 scanAtt008 Mul 40000
9 9:54:07 9:54:39 scanAtt009 Mul 80000
10 9:55:01 9:55:32 scanAtt010 Mul 8000000
11 9:55:33 9:56:04 scanAtt011 Mul 100000
12 9:56:36 9:57:05 scanAtt012 Mul 200000
13 9:57:28 9:57:58 scanAtt013 Mul 400000
scnAtt02_202105131006
10:06:00 10:26:00 Step = 1/2, /4, /6, /8, /10
scnAtt02- 無し

p.70

PL HV 重照射「後」p.72

plhv03-1
2
3
4
5

10:35:33
10:36:39
10:37:49
10:39:37
10:41:17

p.71

10:36:14 plhv03-01 aft scnAtt 500V
10:37:21 plhv03-02 600V
10:39:03 plhv03-03 700V
10:40:52 plhv03-04 800V
10:42:23 plhv03-05 900V

FC=8enA

PL1= 1kcps
0:00:41
0:00:42
0:01:14 少し劣化見える
0:01:150:01:06

scnEDssd： Ecal用
scnEDssd01_202105131104
11:04:00 11:32:00 30s/点scnEDssd011 11:04:25 11:05:07 scanED001 0.0 um
2 11:05:19 11:06:03 scanED002 48.6 um
3 11:06:15 11:06:59 scanED003 100.2 um
4 11:07:11 11:07:55 scanED004 148.8 um
5 11:08:07 11:08:51 scanED005 196.4 um
6 11:09:03 11:09:49 scanED006 249.6 um
7 11:10:01 11:10:45 scanED007 297.2 um
8 11:10:57 11:11:41 scanED008 345.8 um
9 11:11:53 11:12:37 scanED009 397.4 um
10 11:12:49 11:13:33 scanED010 446.0 um
11 11:13:45 11:14:29 scanED011 498.8 um
12 11:14:41 11:15:25 scanED012 547.3 um

p.72
0:28:00

PL 4.5K, Sor 300cps, PandS

0:00:42
0:00:44
0:00:44
0:00:44
0:00:44
0:00:46
0:00:44
0:00:44
0:00:44
0:00:44
0:00:44
0:00:44

1 / 2 ページ

〇SSD@Air2

③整理_RunLog_2105Kr_220204.xlsx1_RunLog2105Kr
Run Summary Sheet : CAMAC & LabView
AttFname
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
21/05/13
21/05/13
21/05/14
21/05/14
21/05/15
21/05/15
以上。

start11:15:37
11:16:35
11:17:31
11:18:27
11:19:23
11:20:19
11:21:15
11:22:13
11:23:09
11:24:05
11:25:03
11:25:59
11:26:55
11:27:51
11:28:49
11:29:45
11:30:41
11:31:37

stop Header
11:16:23 scanED013 596.4 um
11:17:19 scanED014 647.6 um
11:18:15 scanED015 699.8 um
11:19:11 scanED016 723.6 um
11:20:07 scanED017 730.9 um
11:21:03 scanED018 748.4 um
11:22:01 scanED019 753.9 um
11:22:57 scanED020 782.6 um
11:23:53 scanED021 806.4 um
11:24:51 scanED022 844.5 um
11:25:47 scanED023 855.5 um
11:26:43 scanED024 868.3 um
11:27:39 scanED025 878.0 um
11:28:37 scanED026 883.4 um
11:29:33 scanED027 907.2 um
11:30:29 scanED028 932.0 um
11:31:25 scanED029 944.8 um
11:32:21 scanED030 5944.8 um

11:30
14:00
7:20
12:00
0:04
0:04

ビーム調整終了(S04-1)利用スタート
利用終了(ML02-1)利用スタート
利用終了
MT終了

Ender / LogNote
Elapsepg.
0:00:46
0:00:44
0:00:44
0:00:44
0:00:44
0:00:44
0:00:46
0:00:44
0:00:44
0:00:46
0:00:44
0:00:44
0:00:44
0:00:46
0:00:44
0:00:44
0:00:44
0:00:44

コメントp.73
p.73

利用後測定無し

2 / 2 ページ

100 10 5

32TrigWob AuF Kap IC1 PL1 ED

PL2
IC2 Air1 Air2