FS2000

DIGITAL INDICATOR WITH GRAPHIC DISPLAY / TOUCH PANEL(SD CARD SLOT & HIGH SAMPLING RATE)

Performanceupgraded!
Hysterisis Option &Multi-Point Hold Option Added

The best solution for OK/NOK judgment of press fitting and caulking application !!
High responsiveness of 5 kHz to fully utilize the performance of Super Cell !!
A fluctuation of force is shown as a waveform!!
‫ ە‬Two-dimensional OK/NOK judgement can be performed‫ ە‬25000 times/sec. high-speed processing‫ ە‬Variety of interfaces

with a load cell and displacement sensor.
‫ ە‬Analog monitor output

USB / DeviceNet / CC-Link / EtherNet/IP / Ethernet / PROFINET IO

Voltage output is proportionate to the input signal making‫ ە‬4.3-inch color LCD module & touch panel

the recording on recorder convenient.

Operation can be effortlessly performed by a direct touchApprox. 2 V per 1 mV/V strain gauge input

on the touch panel.

Comparison & hold function by waveform display

Improved usability‫ ڦ‬Waveform comparison
This function compares the actual measurement waveform
against the setup High/Low limit waveforms and will give out an
NOK judgment when any of the point exceeded the preset
High/Low limit waveforms.

4.3 inch wide display provides excellent visibility.
Main display configuration can be selected to keep it assimple as possible by eliminating unnecessary information.

‫ ڦ‬Multi-point judgment
OK/NOK judgment can be performed on multi points in one
process. (e.g. The start point and end point of press fitting can be
judged respectively.) (Max. 5 points)<Multi-point judgment>
Peak hold
Inflection point hold+Peak holdPeak(Multiple holds can be adjustedin the same range.)<Waveform
comparison>OK judgment

PeakHi limit setting
Lo limit setting

Inflection point
X axis:time ordisplacement

Y axis: load

Enlarged numerical display

Selectable waveform reference
Judgement is possible based on press point of press machine
Load0

Load

10 mm

Displacement-10 mm0Displacement

Front reference

Back reference

Measurement point starts from left

Measurement point starts from right

Enlarged waveform display

Trend display is helpful for preventive maintenance

Changed setting items are highlighted!

Trend of the zero-point shift and hold values can be monitored
to find any irregularities for preventing breakdown of machines.

Master and current set values are listed up for checking thechanged setting items easily.
Set values can be edited directly on the list without going into
each setting menu.
* Except for waveform comparison settings

Trend display

List display

Saved measured data (waveform) on SD card can be displayed afterwards

Example of use
FS2000Measured data and set values can be saved in
the SD card. Data can be converted to CSV
format easily for editing with Microsoft Excel.

SD cardDisplacement sensor
PCUSB
Load cellI/O

CC-Link, DeviceNet, EtherNet/IP,Ethernet, PROFINET IO

Press machine

PLC

FS2000-HYS Hysteresis specifications
StandardHysteresis specifications

Can see going waveform

Can see outgoing and return waveform‫ ە‬Can choose comparison method!

<Differential waveform comparison>
Judgment of OK/NOK by the difference between
going and returning<Standard waveform comparison>
OK/NOK judgment for going and returning waveform.
kN
40.00

kN
40.00

Going waveform
is OK

Big difference between
going and returning

Returning waveformis NG-10.00-10.000.00

50.00 mm

Recommended for below usage:- Expansion and contraction of the spring- Rotating the steering wheel clockwise,
counterclockwise, etc.

0.00

50.00 mm

Recommended for below usage:- Torque hinges used to open and close doors- Shock absorbers that absorb the impact of tires, etc.

FS2000-MHP Multi hold point specifications

Standard
OK/NOK judgement up to 5 points
Up to 15 point detection
sections can be set

Waveform comparison
function is also available

Multi Hold Point Specifications
2.12.MXGPHQWXSWRSRLQWVSpecifications
Sensorinput(Standard)

Sensor input for load (Fixed as strain gauge input) (6-wire)
Excitation voltage DC 2.5, 5, 10 V±10% (depending on settings) Output current: Within 30mA
Signal input range 㸫2.0 to 㸩2.0 mV/V
AccuracyNon-linearity: Within 0.02% FS±1 digit (at 2.0 mV/V input)
Zero drift:
Within 0.1 μV/Υ RTI
Gain drift:
Within 15 ppm/ΥSelectable from 10 Hz to 10 kHz (-6 dB/oct.) (at A/D converter speed 25000 times/sec.)
Low-pass filter
Selectable from 2 Hz to 2 kHz (-6 dB/oct.) (at A/D converter speed 5000 times/sec.)
A/D converter

Speed:
Resolution:Load digital zero/ Displacement adjustment/ Measurement start/
Measurement end/ HOLD1 to 5/ Reset/
Forcibly light up the backlight/ Touch panel lock/ Work change
Input type: Plus common/ Minus common selectable.
(Minus common is option: [ISC])
To connect a transistor, connect NPN output type (sink type) for
plus common and PNP output type (source type) for minus common.
USB:
USB interface
EIP:
EtherNet/IP interface (option)
ODN:
DeviceNet interface (option)
ETN:
Ethernet interface (option)
CCL:
CC-Link interface (option)
PRT:
PROFINET IO interface (option)(Only one option can be installed)

Interface

Selectable from 25000 times/sec., 5000 times/sec.
24 bit (binary) Effective resolution: Approx. 1/20000 against 2.0 mV/V

Sensor input for displacement (Pulse input: Line driver)Max. input frequency 1 MHz
Internal count range Approx. 1,000,000
Adaptable encoder

Sensorinput
Multisensor
input(Option:[MLT] or[MLT2])

Input signal (16)Output: Incremental type 2-phase output (A/B-phase signal output)
Also capable of single-phase output(A-phase input used. All pulses are counted as in the plus direction.)
Output stage circuit specification: Line driver (Based on RS-422)
Sensor input for load (Strain gauge) (6-wire) … Same as standard
Sensor input for displacement (Pulse input: open collector) ... Other than output circuit, spec is standard [MLT]
Output stage circuit specification: Open collector
Sensor input for displacement (Pulse input: line driver) … Same as standard [MLT2]
Voltage input
Signal input range 㸫10 to 㸩10 VInput impedanceApprox. 1 MΩAccuracyNon-linearity: Within 0.02% FS±1 digit (at 10 V input)
Zero drift:
Within 0.2 mV/Υ RTI
Gain drift:
Within 0.01%/Υ
Low-pass filter
Selectable from 10 Hz to 10 kHz (-6 dB/oct.) (at A/D converter speed 25000 times/sec.)
Selectable from 2 Hz to 2 kHz (-6 dB/oct.) (at A/D converter speed 5000 times/sec.)
A/D converter
Speed:
Selectable from 25000 times/sec., 5000 times/sec.
Resolution: 24 bit (binary) Effective resolution: Approx. 1/20000 against 10 V

ISC: I/O Source board, MLT: Multi sensor input, MLT2: Multi sensor input 2FS2000-HYS:

Special option which records and judge a reverse waveform

FS2000-MHP:

Special opeton which enables to detect hold points up to 15option(Hysteresis Specifications)(Multi Hold Point Specifications)
GeneralPower supply voltage
DC 24 V (±15%)specifications Power consumption
6 W typ.
Operation condition
Temperature:
Operation: 㸫10 to 㸩40Υ
Storage: 㸫20 to 㸩60Υ
Humidity:
85% RH or less (non-condensing)
Dimension
132(W) × 98(H) × 110(D) mm (not including projections)
WeightApprox. 1.0 kg
DeviceNet connector
Attachments I/O connector (with cover) …1(when DeviceNet option is selected) …1
Analog connector…1CC-Link connector
Operating tool…1(when CC-Link option is selected)…1SD card 1 GByte…1Accessories

AnalogOutput level
voltage output
DisplayComp.
&judge.
function

Approx. 2 V per 1 mV/V input
Load resistance: 2 kΩ or more
4.3 inch TFT color LCD module, Display area: 95.0(W) × 53.9(H) mm, Dot configuration: 480 × 272 dot
Display frequency
Fixed at 3 times/sec.
Multi point comparison mode: 16 ch (set values)
Capable of judging up to 5 hold points at the same time.
Sample, Peak, Bottom, P-P, Relative Maximum, Relative Minimum,Inflection Point (A,B,C,D), Average, End displacement
Waveform comparison mode: 16 ch (setting values can be stored)
Compares the actually measured waveform against the preset HI/LO waveforms.
The overall measured waveform will be compared against the preset HI/LO and if anyof its points exceeds the preset waveform, then the measured waveform will be NOK.

OptionSpecialHysteresis
Multi-point comparison and waveform comparison are possible by measuring going/returning
specifications with one waveform. (Can choose go/return difference comparison)
Number of drawing points: 1000 points for going, 1000 points for returning
Multi hold point Multi hold: 15 points
specifications Sampling speed: 5000 Hz
Showing the trend of measurement data to help finding irregularities at early stage.
Preventive
Trend display
maintenance Statistics
supportScreen captureWork name edit
Setting list display
User management
External
Output signal (16)signalUsing the latest 10000 measured data
Displaying number of measurement, OK, NOK
Saves screen capture data as bmp data.
Work name can be edited and displayed for each Work No..
Changed setting items comparing to master set values are highlighted.
Login ID and Password
Point judgment (load, displacement)/ Load overload/
Measurement complete/ Waveform comparison judgment/
Load & displacement OK/ CPU OK/ SD card OK/ Timing output 1,2
Output Type: Sink type/ source type selectable.
(Source Type is option: [ISC])
Output transistor ON at signal ON.
To connect an input unit like a PLC, connect plus
common for sink type, and minus common for source type.
Rated voltage: 30 V, Rated current: 30 mA* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,fluorescent display tubes and LCDs due to manufacturing process or production lots.

Operation manual…1
CN36:
I/O connector (with cover)
CN71:
CC-Link connector
CN72:
Double row connector for CC-Link
CN77:
Analog connector
CND01:
DeviceNet connector

SD1G:
SD2G:
SD16G:
SD32G:
CA81-USB:

SD card 1 GByte
SD card 2 GByte
SD card 16 GByte
SD card 32 GByte
USB cable (A-miniB type) 1.8 m

TSU03:
DC lighting surge unit
CE marking EMC directive EN61326-1
certification

Structure of product code‫ڧ‬ղFS2000ձձ Standard unit‫ڧ‬ճ‫ڧ‬մղ I/O output

մ Interface

SignStandardISCSign
Standard

Output type
Sink type (NPN output)
Source type (PNP output)

InterfaceUSB↓ One optional interface can be added
in addition to the standard interface.

ճ Sensor input

ODN
CCL
EIP
ETN
PRTSign
Output type
Standard Strain gauge,Line driver
MLT orStrain gauge
MLT2Open collector (MLT only)
1*Line driver (MLT2 only)
Voltage (Load or displacement)

DeviceNet
CC-Link
EtherNet/IP
Ethernet *2
PROFINET IO*2 When choose ETN option,
USB interface is not included.

*1 When choose MLT option, ULE-50 is not available to use.

Combination table
X axis

Y axisTimeLine driverLine driver
Time
Open collector
Open collector

Strain gauge
Strain gauge
Voltage (Load)
Voltage (Load)
Strain gauge
Voltage (Load)

Voltage (Displacement)

Strain gauge

Standard

MLT

MLT2○○○○○○○

×××××××

○
○
○
○

××

○

External dimension(Side)(Rear)

Unit: mm98

SENSOR1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 1691

Panel cutout size
DC IN

Panel thickness
1.6 to 3.2 mm

CONTROL
USBMADE IN JAPAN

132

10100

SER. NO.

124

Digital contact sensor ULE-50
Digital contact sensor

ULE-50A digital contact sensor
381A-LDI.
designed for FS2000 and F381A-LDI.
dgmentYou can perform OK/NOK judgmentwith a Force vs Displacement curve.

Wide measuring range & high-accuracy
Measuring range: 50 mm
Resolution: 2.5 μm

125 +1092 +10(Front)