③整理_RunLog_2109C(MS-EXP21-08)_220204.xlsx1_RunLog

2109C-MS2 (MS-EXP21-08)
Run Summary Sheet : CAMAC & LabView

条件全て書くこと
Att

Fname

startstopHeader

実験前Ender / LogNote
Elapsepg.
LogNote: E5exp(8)_コメント

100 10 532Trig

Wob AuF Kap IC1 PL1 ED

Det.Setup
PL2 Air1 Air2 PL2,IC2,SSD
IC2

12C6+ 135MeV/u 32.6MHz Hyper-ECR 1pnA
befsA2: SAmp燃えた！ SA交換した
p.8
21/09/27 1_old 15:04:00 15:17:22
sB1 SA kosyou?
前回CMS1のママで再確認
21/09/28
1 8:49:15 9:33:16
sB1 Ok dattaSAmp交換後、SSD 2107C-MS1:Ar-gain のママで確認
2 9:34:09 9:58:03 bef-02 SampGain x5
pul0.3 0.5 1. 2. 3. 4. SAmp gain 一律５倍
sA1:x0.1x100x5.0(8dB), sA2:x0.1x100x6.5(2+8)
3 17:13:31 17:33:10 bef-03(Fin) SAmpx5 + sB2:16->4dB pul 0.3 0.5 1. 2. 3. 4. C用決定
sB1:x0.1x100x8.0(1+4dB), sB2:x20x6.0(4)plBGp.12
PL beam OFF 確認
21/09/29
1 19:09:20 9:47:03 plBG01 PL1_3000um 900V PL2_500u 900V
C.Ray mienai
21/09/30
2 13:08:35 14:21:58 PLGB02- IzumiRun 200uSv/h
end PL2 mo Trg ni ireta
3 17:46:43 17:48:22 plBG03 Gaf06 PL1,2,900V
end 2min
実験中：Beam調整時
21/09/30
13:40:00
泉Grp終了、電源切替E5B->A ～1hr だった
p.18
ZnSスポット調整Cap
1 16:40:00
R01_ZnS_spot
mpg
×50u ×
1 0 0 0 4
2
R02_TQ中1A振り
Wob=C135で3
R03_Wob R40-10
R=40,35,30,..,10
1 0 0 0 2
p.18
Gaf:: EBT3 □85x85mm @ Ysld
Gaf焼き(1)Gaf
1 17:15:00
Au 112.5u本命, R=090s
1 1 1 0 42
Au 112.5u, R=045s
3Au 112.5u, R=0180s
4Au 47u, R=0120s
5Au 47u, R=06min
6Au 47u, R=04min
11 17:50:00
BmSpot10s
1 1 1 0 412
Au 65.5u, R=0120s
13AL 1mm,R=090s
14
Au 112.5u, R=4040s
15
Au 112.5u, R=3040s
16
Au 112.5u, R=2040s
p.19
回路調整
18:00:00
PL1=3mm(2107CMS1使用) PL2=500u(2105Ar使用)
PL1 Thr=125, PL2=134mV 前のママ
PL1=PL2= 250cps
0 2 0 0 4 PorS(0) R35 112u
〇PL HV（１：利用「前」）plhv011 19:03:07 19:04:12 plHV01-01 700V
0:01:05 PL1=800V決定2 19:04:44 19:05:48 plHV01-02 800V
0:01:04
3 19:06:21 19:07:15 plHV01-03 900V
0:00:54
4 19:07:44 19:08:59 plHV01-04 1000V
0:01:15××145 200

3mm 0u

500u

0u

IC2大fix無し

scnAtt： IC1 vs PL1
scnAtt_202109302001
20:01:00 20:09:00 Beam増やす方向で

p.20

神原測定
20:25:00 20:30:00 ED出口コリt3cm有り 5min照射GV5A1 close で開かないp.20

0:08:00

1 1 1 0 3 ～1 0 0 0 1

？ED=128B=18422.8um
0:05:00
FC5A1=6enA, IC1=1-6Aくらい

0 1 0 0 2

Att急激に変化。うまい組合せ無い。

〇〇

IC2@Air2,X=0電磁弁ネジ調整で復帰。週番にtel。３０分延長許可。

scnEDic： ExpR 確認
scnEDic_202109302104
30sec
21:04:00 21:32:00 本番p.20

plhv02-1234

21:49:26
21:50:39
21:51:52
21:53:01

IC1～IC2= 2.7-8A

var

IC2大

0 2 0 0 1 PorS(0)3mm 0u500u0 2 0 0 1 PandS

var

1 0 0 0 2

0:28:00

PL HV（２：重照射「後」）PL1=PL2= 1kcps
0:00:43 劣化少ない
0:00:45
0:00:40
0:01:32

21:50:09 plHV02-01 700V
21:51:24 plHV02-02 800V
21:52:32 plHV02-03 900V
21:54:33 plHV02-04 1000V

scnEDssd： C-gain でタメ用scnEDssd01_202107012148
21:48:00 22:32:00 20sec/点
scnEDssd011 22:21:30 22:21:49 scanED001 0.0 um
2 22:22:00 22:22:21 scanED002 2890.0 um
3 22:22:32 22:22:51 scanED003 5194.0 um
4 22:23:02 22:23:21 scanED004 8084.0 um
5 22:23:32 22:23:53 scanED005 10314.0 um
6 22:24:04 22:24:24 scanED006 13204.0 um
7 22:24:36 22:24:57 scanED007 14904.5 um
8 22:25:08 22:25:27 scanED008 15508.0 um
9 22:25:38 22:25:56 scanED009 16484.2 um
10 22:26:08 22:26:27 scanED010 16964.6 um
11 22:26:38 22:26:58 scanED011 17154.2 um
12 22:27:10 22:27:29 scanED012 17283.8 um
13 22:27:40 22:27:59 scanED013 18398.0 um
14 22:28:10 22:28:29 scanED014 18702.3 um
15 22:28:40 22:28:59 scanED015 18878.5 um
16 22:29:10 22:29:29 scanED016 18963.8 um
17 22:29:40 22:29:59 scanED017 19028.8 um
18 22:30:10 22:30:29 scanED018 19068.0 um
19 22:30:40 22:30:59 scanED019 19122.3 um
20 22:31:10 22:31:29 scanED020 19172.9 um
21 22:31:41 22:32:01 scanED021 19197.7 um
22 22:32:14 22:32:33 scanED022 19222.1 um
21/09/30
22:32:00
MT終了
以上。

p.22
0:44:00

PL ?K, Sor 1K cps

0:00:19
0:00:210:00:19
0:00:19
0:00:210:00:20
0:00:210:00:19
0:00:18
0:00:19
0:00:20
0:00:19
0:00:19
0:00:19
0:00:19
0:00:19
0:00:19
0:00:19
0:00:19
0:00:19
0:00:20
0:00:191 / 1 ページ

SSD@Air2