10L ROTARY EVAPORATOR
Elaborate and space-saving design
with versatile ability

10L Standard
Model N-3100
5LSample flask
diameter
φ100㎜

Replaceable with
optional 5L sample flask

Bath heater capacity
3.5kW
Temperature control range
RT＋10 to 180℃
(Bath for water or oil)

Built-invacuum controller

Rotation and temperature
controller

Vacuum controller
NVC

Power

Leak

Vacuum controller, rotation and temperature control parts
are newly designed. The liquid crystal display (LCD) isre-designed and enlarged. It allows individual function
information to be displayed and is easier to read and operate.

N-3100Enhanced safety features

Float switch detects low water level in the bath and prevents
empty boil

Safety cover to prevent hands and clothes from gettingcaught during rotation

TOKYO RIKAKIKAI CO., LTD.

greatly to reduces working

ModelCat. No.
Sample flask
Rotation speed range
Evaporation capacity
Ultimate vacuum
Temperature control range
Temperature control accuracy

N-3100280748

10L standard system

System code SYS09483

10L Round shaped flask (Diameter 100mm)
10 - 140 rpm (Continuous variable speed)
Max. 3.5L/h (Water)
2.6hPa (2Torr)or lessRoom temp. +10 to 180℃ (Water or oil)

±1℃/±3℃ (with water/oil, when flask rotating)

Vacuum control

Controls (Constant value, Program, Stepprogram, auto [temperature sensor is optional]

Receiving flask

5L round shaped flask (with drain valve)

Bath dimension, capacity,and material

ID360x200Hmm, 20L, SUS304

Connection diameter

Suction/cooling water nozzle OD 16mm
Control solenoid valve nozzle OD 13mm

Outer dimensions(mm)/weight

930W×450D×1890H /Approx. 103kg

Power input and supply voltage

17A, 3.9kVA/230V single phase 50/60HzPower utility

230VAC single phase 23A capacity

This system includes the following products.
※Specifications are room temperature 20℃, rated power supply voltage,
50Hz, and no load.
※Evaporation capacity varies depending on conditions such as degreeof vacuum, bath temperature, and cooling water temperature.
※Temperature control range depends on the ambient temperature andevaporation rate.
※Power plug is not included.

Product name
Model
N-3100Rotary Evaporator
CA-2610
Low temp. circulator
N950
Diaphragm vacuum pump
光ファイバーFor N950
Power cord
センサ
Cooling hose ID15mm 5mx2 pcs
Vacuum hose ID9xOD21mm

Cat. No.
280748270140251170245370174460119190

Options (for medium to large size rotary evaporators)
Sample injection/solvent discharge unit
Model IDU-1000 (Cat. No. 259820)
IDU-1000 is used in combination with themedium to large size rotary evaporators.
The fiber-optic sensors installed in thesample flask and receiving flask detectincrease or decrease in sample and solvent
volume, allowing continuous injection ofsample into the sample flask and dischargeof solvent from the receiving flask (PAT.)
without stopping the system(withoutreleasing the vacuum). Thus, IDU-1000
greatly reduces working time.

Fiber opticsensor

Capillaryforinjection

・Ultimate vacuum: Vacuum in rotary
evaporators + 3hPa
・Solvent discharge pump:
Max. volume 0.8/0.96L/min(50/60Hz)
・Operating fluid temperature: 10 to 40℃※ Vessels for sample injection and solvent
discharge are not included.
※ Acid-based samples cannot be used.

Sample flask

Example of model IDU-1000
configuration with 20L
rotary evaporator
Solenoid valve(for sample injection)
Capillary for injection
Fiber optic sensor

Sampleinjection

Fiber optic
sensorReceivingflask
Solvent
discharge

From sample injection
container
Control box
Leak vacuum hose
Leak nozzle(for solvent discharge)
Fiber optic sensor
Leak prevention sensor
Receiving flask (5L)
Receiving flask support
To Solvent discharge
container
Vane pump

See the video
詳細動画は、
on
N-3100こちら

Liquid level control unit
Cat. No. 279860
Option to stop sample injection/concentration
by detecting the amount of sample in thesample injection vessel/amount of solventin the discharge solvent vessel by a fiberoptic sensor.

TOKYO RIKAKIKAI CO., LTD.

e.

https://eyelaworld.com

Fiber optic sensor (1pc, 2.5m)
Cat. No.
Teflon® coating279710
0.5m
2797201m※Two fiber-optic sensors are required when using the liquid levelcontrol unit.

TN Koishikawa Bldg.
1-15-17 Koishikawa
Bunkyo-ku, Tokyo
112-0002 Japan

Tel: +81-3-6757-3378
Fax: +81-3-3868-6571
E-mail: contact@eyelaworld.com※The appearance and specifications of the products are subject
to change without notice.