ReminderThanks for purchasing the product. The manual should be in safekeeping
and must be fully understood prior to any usage. The user manual is not aquality certificate. It has some typographical errors. It does not refer toimprovements to the product. The reprinted user manual will include therevised version without any prior notice.
① Package contents
One power meter. One piece of the user manual.
② Product introduction
・HLP‐200B is a hand-held power meter, especially designed to measuresuper low power CO2 laser tube. It is small, portable, and uses an internalbutton battery. It is also convenient to use, easy to read, and has highaccuracy.
Product name: Hand-held power meter
Model: HLP-200B
Measuring scope: 0-200W
Aligned wavelength: 10.6
Battery: 3v/3.6v 2032 button battery③ Upgraded functions
・ HLP-200B is the upgrade version of HLP-200.

It still maintains thefeature of super low power consumption with an improved way of working.
It stands by as soon as it turns on, and begins measuring power when thelaser is emitted to the sensor.
・To achieve better accuracy, this upgrade version has a smaller sensor area.
・Another screw hole (M4) is added so that the power meter can be attached
to a holder that further decreases the deviation due to crooked positions whilein operation.
・To make the replacement of battery more convenient, the way of opening
the battery cover has changed. So users can replace the battery without anytools.
・Status instruction codes are added:“----” blinks represent the power meter is standing by.
“HOT” shows the power meter is over heated.
“Err” means the measured power is beyond the measuring scope of the meter.

1. Target surface

2. Switch

3. Display

4. Battery cover

5. M4 screw hole

④ Operation instructions
・Press the blue button to turn the power meter on and the screen shows“----” with a beep. Three seconds later, the screen starts blinking “----”, whichmeans the power meter is in a standby mode. Within three seconds after
turning it on, laser should be emitted for six seconds continuously. Ensure
the measured laser is emitted to the center of the sensor. The power meter
will begin measuring when the sensor receives laser. When the “----” stops
blinking on the screen with a beep, the measurement has started.
・By the end of the measurement, the screen shows the measured power witha beep. It stays for six seconds. The equipment will shut down withanother beep.
・When the screen shows “HOT” with a beep, the sensor is over heated. Itis required to cool down prior to another measurement.
・When the screen shows “Err” with a beep, the measured power is beyond
the measuring scope of this power meter.
・Use just fingers to remove the battery cover to change batteries.

⑤ Reminder・Handle with great care in case you damage the precision equipment.
・Ensure the laser is emitted to the center of the sensor!
・It is suggested to cool the power meter down to the ambient temperatureeach time if you repeat measurements.
⑥ After-sales service
The product comes with one year warranty with lifetime maintenance. The
warranty will be invalid under situations below.
・Out of the warranty period.
・Force majeure.
・Failures or damage due to unauthorized uninstallations, modifications, orrepairs.