③整理_RunLog_2201Kr_220201.xlsx1_RunLog

2201KrRun Summary Sheet : CAMAC & LabView

条件全て書くこと
Att

Fname

start

stop

Header

実験前

Ender / LogNote
Elapsepg.

コメント

100 10 5

32TrigWob AuF Kap IC1 PL1 ED

Det.SetupAir1 Air2 PL2,IC2,SSD

×145 200

LogNote: E5exp(8)_

実験中：Beam調整時
22/01/09
Cap

ロス0.5hZnSスポット調整

p.112
Cap01_上流中で下流は右Cap02_ステアラで下流中心へ
Cap03_調整中 右上切れるのは何故
Cap04_調整後 右上切れたまま
Cap05_Zslide 軸はOK
Cap06_Z71でQdef
7:10:00 右上切れ調整後、上流Trans再確認
7 7:35:00
Cap07_WobR=0,36,0
8Cap08_WobR=0,Au48IN
p.113
Gaf:: EBT3 □85x85mm @ Ysld
Gaf焼き(1)1234566:24:00

Att vs IC2@Air2Gaf

1234568:00:00

9:15:00☓plhv02☓
ロス2h
plhv03-

1234
1234
1234

9:48:38
9:50:25
9:51:35
9:52:48
10:27:57
10:29:32
10:30:42
10:32:34
10:54:40
10:55:49
10:57:07
10:58:29

0 2 0 1 1×50u ×××

右側が切れている？
var: WobR =12C 135MeV/u

R=0,36,00 2 0 1 00 2 0 0 0Rvar
OFF 48u
×0 4 0 0 1×0u

IC2@Xclmt

OFF 48uOFF 73u
OFF 2.5u
OFF 5uR36 48u
R20 48u

60sec 薄い
60sec 薄い
7sec
55sec 間違ってY=+22もやった
90sec
90sec

PL1,2 Dscr: 2110Arのママ PL1=125, 899mV, PL2=134mV

p.114
PL1=100u#2(new) PL2=500u#2(new)PL HV（１：利用「前」）plhv01-mpg
mpg
IC2[A]
8.3-9
6.4-9
1.3-8
1.4-8
4.8-9
7.4-9

Au 48u, R=0
Au 73u, R=0
Au 2.5u, R=0
Au 5u, R=0
Au 48u, R=36
Au 48u, R=20

回路調整

png
png
mpg
png
mpg
mpg

9:49:31 plhv01-01 700V PL1warui
9:51:04 plhv01-02 800V
9:52:11 plhv01-03 900V
9:53:45 plhv01-04 1000V
10:28:37 plhv02-01 700V PL100um#1 nishita
10:30:08 plhv02-02 800V
10:31:43 plhv02-03 900V
10:33:28 plhv02-04 1000V
10:55:17 plhv03-01 700V PL1#1 syakou
10:56:42 plhv03-02 800V
10:58:02 plhv03-03 900V
10:59:10 plhv03-04 1000V

PL1=PL2=1kcps
2 2 0 0 0 PorS(0) R36 48u
end OK same 0:00:53
as 2111Kr
PL1=100u#2変？PH低い
e 60sec PL1nashi
0:00:39e PL1 girigiri 0:00:36
e PL1 yatto 0:00:57
e PL1 200ch Habahiro0:00:40 PL1=100u#2 --> 100u#1(2111Kr使用)に交換
e0:00:36 でも光漏れアリe PL1 mienai?0:01:01
e0:00:54
e PL1 PH henka
0:00:37nashi?
PL1=100u#1 800Vでやる
e modotta 0:00:53
e0:00:55
e0:00:41〇

100u 0up.115

scnEDic： ExpR 確認
scnEDic01_202201091127
30sec/点☓ 11:27:00 11:44:00
scnEDic02_202201091213
30sec/点ロス1h☓ 12:13:00 12:39:00
scnEDic03_202201091245
30sec/点12:45:00 13:23:00

1 1 0 0 0R36 48u〇100u var

PL2=inのまま
IC2大
PL2=inのまま
IC2大
PL2=inのまま
IC2大

1 1 0 0 0 ～1 3 0 0 3

R36 48u〇100u 0u

IC2@Air2,X=0

PL1=3kcps
0:01:15 PL1=100u#1 光漏れ修理した
0:00:35
0:00:48
0:01:00

2 2 0 0 0 PorS(0) R36 48u〇100u

0:47:00 PL 6K, Sor 400cps, PandS

2 1 1 0 0 PandS R36 50u

IC2大 100u var

IC1～IC2=6.1-8A
0:17:00 ん？200μ厚い？

p.116
0:26:00 ExpR=700um付近パラメータでIC1出口カバーが入っていた
0:38:00

scnAtt： IC1 vs PL1
scnAtt01_202201091336
13:36:00
scnAtt011 13:37:26
2 13:38:23
3 13:39:19
4 13:40:27
5 13:41:43
6 13:42:51
7 13:44:03
8 13:44:59
9 13:45:51
10 13:46:24
11 13:47:19
12 13:48:19
13 13:49:11
13:55
22/01/09
16:20
22/01/10
4:00

p.117
13:49:00 Step = 1/2, /4, /6, /8, /10

0:13:00 scnAtt本番

13:37:56 scanAtt001 Mul 1000
13:38:54 scanAtt002 Mul 2000
13:39:48 scanAtt003 Mul 4000
13:40:58 scanAtt004 Mul 8000
13:42:12 scanAtt005 Mul 10000
13:43:22 scanAtt006 Mul 20000
13:44:34 scanAtt007 Mul 40000
13:45:30 scanAtt008 Mul 80000
13:46:22 scanAtt009 Mul 800000
13:46:56 scanAtt010 Mul 100000
13:47:48 scanAtt011 Mul 200000
13:48:48 scanAtt012 Mul 400000
13:49:40 scanAtt013 Mul 800000
ビーム調整終了(ML03-1)利用スタート(ML02-04)利用終了

0:00:30 1/f=1/999～1/9 手動で変更
0:00:31
0:00:29
0:00:31
0:00:29
0:00:31
0:00:31
0:00:31
0:00:31
0:00:32
0:00:29
0:00:29
0:00:29
辛うじて14:00までに終了：今回はロスタイムが多かった

PL HV（２：光漏れ修理）
plhv04-1
2
3
4

6:19:53
6:21:33
6:22:35
6:23:48

p.118
p.129

6:21:08 plhv04-01 700V ML03aft
6:22:08 plhv04-02 800V
6:23:23 plhv04-03 900V
6:24:48 plhv04-04 1000V

scnEDssd： Eスペクトル(１) Ecal用
scnEDssd01_202201100637
6:37:00 7:24:00 Ecal用: 39点x測定30sec
scnEDssd011 6:37:53 6:38:49 scanED001 0.0 um
2 6:39:01 6:40:01 scanED002 48.6 um
3 6:40:13 6:41:13 scanED003 100.2 um
4 6:41:25 6:42:25 scanED004 148.8 um
5 6:42:37 6:43:38 scanED005 197.4 um
6 6:43:50 6:44:51 scanED006 249.5 um
7 6:45:03 6:46:03 scanED007 297.5 um
8 6:46:15 6:47:15 scanED008 345.8 um
9 6:47:27 6:48:27 scanED009 397.4 um
10 6:48:39 6:49:39 scanED010 446.0 um
11 6:49:51 6:50:51 scanED011 498.8 um
12 6:51:03 6:52:03 scanED012 547.3 um
13 6:52:15 6:53:15 scanED013 596.4 um
14 6:53:27 6:54:27 scanED014 647.6 um
15 6:54:39 6:55:39 scanED015 699.6 um
16 6:55:51 6:56:51 scanED016 724.8 um
17 6:57:03 6:58:03 scanED017 748.2 um
18 6:58:15 6:59:15 scanED018 756.0 um
19 6:59:27 7:00:27 scanED019 760.6 um
20 7:00:39 7:01:39 scanED020 770.8 um
21 7:01:51 7:02:51 scanED021 777.9 um
22 7:03:03 7:04:03 scanED022 783.1 um
23 7:04:15 7:05:15 scanED023 792.8 um
24 7:05:27 7:06:27 scanED024 807.6 um
25 7:06:39 7:07:39 scanED025 817.8 um
26 7:07:51 7:08:51 scanED026 830.6 um
27 7:09:03 7:10:03 scanED027 844.5 um
28 7:10:15 7:11:15 scanED028 858.7 um
29 7:11:27 7:12:27 scanED029 869.5 um
30 7:12:39 7:13:39 scanED030 879.8 um
31 7:13:51 7:14:51 scanED031 887.9 um

p.130
0:00:56
0:01:00
0:01:00
0:01:00
0:01:01
0:01:01
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00

1 / 2 ページ

SSD@Air2

③整理_RunLog_2201Kr_220201.xlsx1_RunLog

2201KrRun Summary Sheet : CAMAC & LabView

条件全て書くこと
Att

Fname
32
33
34
35
36
37
22/01/10
22/01/11

start
7:15:03
7:16:15
7:17:27
7:18:39
7:19:51
7:21:03
11:30
10:05

stop Header
7:16:03 scanED032 898.1 um
7:17:15 scanED033 908.4 um
7:18:27 scanED034 918.6 um
7:19:39 scanED035 921.2 um
7:20:51 scanED036 925.7 um
7:22:03 scanED037 944.8 um(S05-1)利用スタート(S05-1)利用終了

PL HV（３：利用途中）
plhv05-

plhv06-22/01/11
22/01/12

1
2
3
4
1
2
3
4

11:17:05
11:18:40
11:20:20
11:24:05
12:02:11
12:05:01
12:05:56
12:07:41
12:09:35
16:50
4:30

11:17:44 plhv05-01 700V aft S05
11:19:53 plhv05-02 800V
11:22:57 plhv05-03 900V
11:25:36 plhv05-04 1000V
12:02:45 plhv06-01 PL100u#2 800V12:05:33 plhv06-02 900V
12:06:37 plhv06-03 1000V
12:08:38 plhv06-04 1100V
12:10:23 plhv06-05 1200V(R02-2)利用スタート(R02-2)利用終了

Ender / LogNote
Elapsepg.
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
0:01:00
p.131

p.133
6:25:00 5mmStep x 30sec
p.134
7:11:00 5mmStep x 30sec
p.135
7:35:00 5mmStep x 30sec

scnZｓｓｄ Krで練習
scnZssd01_202201120754
7:54:00 8:10:00 25mmStep x 120sec
scnZssd011 7:54:15 7:56:11 scanZ001 0.0 mm
2 7:56:14 7:58:15 scanZ002 25.0 mm
3 7:58:18 8:00:16 scanZ003 50.0 mm
4 8:00:20 8:02:20 scanZ004 75.0 mm
5 8:02:24 8:04:23 scanZ005 100.0 mm
6 8:04:26 8:06:27 scanZ006 125.0 mm
7 8:06:30 8:08:31 scanZ007 150.0 mm
8 8:08:34 8:10:34 scanZ008 175.0 mm
9 8:10:38 19:44:31 scanZ009 200.0 mm
22/01/12
8:10:00
MT終了
8:50:00 E(Kr) = 69.2 MeV/u

32p.136

IC1=IC2=4.8-8A PL1=OFF1 1 0 0 0
ED=234678=869.6um
0:18:00 失敗。先ず手動でscnZすべきだった。
ED=245678=944.8um
0:23:00 SRIM計算値と合わない！
ED=45678=932.0um
0:21:00 12.8um抜いたので 35mmズレる筈
PL1=6k, Sor=350cps
2 1 0 0 2
ED=234678=869.6um
0:16:00 失敗。先ず手動でscnZすべきだった。
0:01:56
0:02:01
0:01:58
0:02:00
0:01:59
0:02:01
0:02:01
0:02:00 ８点目でSSD来ないヤメ。
11:33:53

加速器E測定： タマキ、小山
8:25:00

100 10 5Trig

Wob AuF Kap IC1 PL1 ED

p.131
PL1=3kcps
2 2 0 0 0 PorS(0) R36 48ue PL100um#10:00:39
nomi PL1=100u#2 まだ > Thre
0:01:13e
0:02:37e
0:01:31e PH hikui 0:00:34 PL1=100u#1 光漏れ修理したが、、e0:00:32 まだ何か変！ PMT or HVコネクタ が悪い？
e
0:00:41 今回は使わないe0:00:57
ep.132scnZic Krで練習
scnZic01_202201120607
6:07:00
scnZic02_202201120648
6:48:00
scnZic03_202201120714
7:14:00

コメントp.137

以上。

2 / 2 ページ

〇100u

R36 48u〇

100u

R36 48u〇100u

Det.SetupAir1 Air2 PL2,IC2,SSD