Agilent 11899AProbe Power Supply
Technical Specifications

The Agilent 11899A probe power supply provides
regulated DC power to a number of Agilent products.
Two probe power outputs are available. Functionally, theAgilent 11899A is equivalent to the probe power
supplied on the front panels of many Agilent spectrum,
network, and lightwave component analyzers.

Compatible products
Agilent 85024A
High Frequency Probe, 300 kHz to 3 GHz
Agilent 41800A
Active Probe, 5 Hz to 500 MHz
Agilent 83400 to 83412 Lightwave Source and Receivers

Typical performance characteristics
Power Output:Power Input:

Dimensions:
Weight:–12.6 V ±6%, 350 mA and +15 V ±6%, 500 mA(male connectors)
88 to 112 V, 50/60 Hz option 100
105 to 135 V, 50/60 Hz option 120
193 to 247 V, 50/60 Hz option 220
211 to 269 V, 50/60 Hz option 240
81 mm wide, 65 mm high, 121 mm long
(3.2 in x 2.6 in x 4.8 in)
net 0.9 kg (2 lb); shipping 1.8 kg (4 lb)

Ordering information
Agilent 11899A
Option 11899A-100
Option 11899A-120
Option 11899A-220
Option 11899A-240

Probe Power Supply(must order one option)
100 V operation
120 V operation220 V operation
240 V operationwww.agilent.com/find/emailupdates
Get the latest information on theproducts and applications you select.

Agilent Technologies’
Test and Measurement Support, Services, and AssistanceAgilent Technologies aims to maximize the value you receive, while minimizing your risk andproblems. We strive to ensure that you get the test and measurement capabilities you paid for andobtain the support you need. Our extensive support resources and services can help you choose theright Agilent products for your applications and apply them successfully. Every instrument andsystem we sell has a global warranty. Support is available for at least five years beyond theproduction life of the product. Two concepts underlie Agilent’s overall support policy: “OurPromise” and “Your Advantage.”

Our Promise
Our Promise means your Agilent test and measurement equipment will meet its advertised
performance and functionality. When you are choosing new equipment, we will help you withproduct information, including realistic performance specifications and practical recommendations
from experienced test engineers. When you use Agilent equipment, we can verify that it works
properly, help with product operation, and provide basic measurement assistance for the use ofspecified capabilities, at no extra cost upon request. Many self-help tools are available.

Your Advantage
Your Advantage means that Agilent offers a wide range of additional expert test and measurement
services, which you can purchase according to your unique technical and business needs. Solveproblems efficiently and gain a competitive edge by contracting with us for calibration, extra-cost
upgrades, out-of-warranty repairs, and on-site education and training, as well as design, system
integration, project management, and other professional engineering services. Experienced Agilent
engineers and technicians worldwide can help you maximize your productivity, optimize the returnon investment of your Agilent instruments and systems, and obtain dependable measurement
accuracy for the life of those products.

By internet, phone, or fax, get assistance with all your test & measurement needs.
Online assistance:

www.agilent.com/comms/lightwave
Phone or Fax
United States:(tel) 1 800 452 4844
Canada:(tel) 1 877 894 4414
(fax) (905) 282 6495
China:(tel) 800-810-0189(fax) 1-0800-650-0121
Europe:(tel) (31 20) 547 2323
(fax) (31 20) 547 2390
Japan:(tel) (81) 426 56 7832
(fax) (81) 426 56 7840
Korea:(tel) (82-2) 2004-5004(fax)(82-2) 2004-5115
Latin America:(tel) (305) 269 7500
(fax) (305) 269 7599
Taiwan:(tel) 080-004-7866(fax) (886-2) 2545-6723
Other Asia Pacific Countries:(tel) (65) 375-8100(fax) (65) 836-0252
Email: tm_asia@agilent.com
Product specifications and descriptions in this document subject to change without notice.
© 1991, 2002 Agilent Technologies, Inc.
Printed in USA April 22, 2002
5091-0754E