③整理_RunLog_2107C(MS-EXP21-07)_220203.xlsx1_RunLog

2107C-MS1 (MS-EXP21-07)
Run Summary Sheet : CAMAC & LabView

条件全て書くこと
Att

Fname

startstopHeader

実験前Ender / LogNote
Elapsepg.
LogNote: E5exp(7)_100 10 5

32TrigWob AuF Kap IC1 PL1 ED

Det.Setup
PL2 Air1 Air2 PL2,IC2,SSD
IC212C6+ 135MeV/u 32.6MHz Hyper-ECR 1pnA

bef21/06/29

1 16:15:21 16:40:52 SSD 2105 Ar-gain のママ
p.108
実験中：Beam調整時
21/07/01
13:30:00
泉Grp終了、電源切替E5B->A ～２hr 必要
p.141
ZnSスポット調整Cap
1 15:56:00
R01_ZnS_Qmag, Z=0-300
2
R02_ZnS@Au Focus
3
R03_AZDハング
4
R04_ZnS@Au=out, WobR=10,20,30,40 LightON
5
R05_ZnS@Au=out, Light=OFF, WobR=40,35,-,10
6
R06_ZnS@AuなしでWobON7
R07_WobOFF,ZnS_out, Au20um_in
8
R08_Gaf01_Au20um
9
R09_Gaf02oe03
10R10_Gaf04_Au121um
11
R11_Gaf06_出口コリ
p.142
Gaf:: EBT3 □85x85mm @ Ysld
Gaf焼き(1)Gaf
1 17:30:00
Au 20u, R=02
Au 20u, R=03
Au 49u, R=04
Au 112u本命, R=05
Au 112u, R=35(C135)
6同上＋出口コリ□30x30
p.142
回路調整
18:00:00
PL1=3mm(new) PL2=500u(2105Ar使用)

PL HV（１：利用「前」）plhv01-コメント1234

18:29:06
18:30:36
18:31:58
18:33:40

18:30:10 plhv01-01 700V
18:31:33 plhv01-02 800V
18:33:09 plhv01-03 900V
18:34:50 plhv01-04 1000V

IC1-HV プラトー確認： 手動(new)18:55:00

mpg

0 2 1 0 3

Wob=C135で×50u ×××145 200

0 2 1 0 20 2 1 0 15s
25s
45s
60s
60s
60s

0 2 1 0 4
1 1 1 0 4

PL1 Thr=309, Low900mV
PL1=PL2= 400cps
0:01:04 PL1=800V決定0:00:57
0:01:11
0:01:10p.143

3 1 1 0 3 PorS(0) R35 112u

〇3mm 0u500uvar

IC2大3mm 0u500u

1 2 0 0 2
電流値はほぼフラット -> 通常値 200V でやる

20～200(通常)～350V

scnEDic： ExpR 確認
scnEDic01_202107011910
30sec
19:10:00 19:23:00 ラフスキャン
scnEDic02_202107011924
30sec
19:24:00 19:56:00 本番 44/51点

p.143

1234

20:05:59
20:07:15
20:10:44
20:12:28

0 2 1 0 2

p.144
0:32:00

PL HV（２：重照射「後」）plhv02-

IC1～IC2= 1.0-7A
0:13:00

PL1=PL2= 400cps
0:00:49 劣化少ない
0:02:16
0:01:160:02:30

20:06:48 plhv02-01 700V
20:09:31 plhv02-02 800V
20:12:00 plhv02-03 900V
20:14:58 plhv02-04 1000V

3 1 0 0 2 PorS(0)scnAtt： IC1 vs PL1
scnAtt01_202107012039
20:39:00
scnAtt02_202107012049
20:49:00
scnAtt03_202107012105
21:05:00
scnAtt031 21:06:31
2 21:07:26
3 21:07:57
4 21:08:50
5 21:10:14
6 21:11:18
7 21:12:10
8 21:13:27
9 21:14:19
10 21:14:51
11 21:15:46
12 21:16:50
13 21:17:54
14 21:18:29

p.142

1 1 0 0 3 ～3 1 0 0 2

20:47:00 ビーム減らす方向で

0:08:0021:00:00 ビーム増やす方向で0:11:00

21:18:00 十分下 100cps から増やす方向で0:13:00

21:07:00 scanAtt001 Mul 40000000
21:07:56 scanAtt002 Mul 400000000
21:08:27 scanAtt003 Mul 20000000
21:09:22 scanAtt004 Mul 8000000
21:10:46 scanAtt005 Mul 4000000
21:11:50 scanAtt006 Mul 2000000
21:12:40 scanAtt007 Mul 800000
21:13:58 scanAtt008 Mul 400000
21:14:50 scanAtt009 Mul 20000000
21:15:22 scanAtt010 Mul 200000
21:16:18 scanAtt011 Mul 80000
21:17:21 scanAtt012 Mul 40000
21:18:28 scanAtt013 Mul 200000
21:19:02 scanAtt014 Mul 20000

0:00:29 1/f=1/999～1/9 手動で変更
0:00:30
0:00:30
0:00:32
0:00:32
0:00:32
0:00:30
0:00:31
0:00:31
0:00:31
0:00:32
0:00:31
0:00:34
0:00:33
ED=128B=18422.8um
0:05:00
FC5A1=9enA, IC1=1.09-6A

0 1 1 0 2

0:16:00

2 2 0 0 4 PandS

神原測定
21:35:00 21:40:00 ED出口コリ無し 5min照射

p.147

IC2@Air2,X=0〇

〇fix

無し

scnEDssd： ED厚さ較正用scnEDssd01_202107012148
21:48:00 22:04:00 3sec/点
scnEDssd011 21:48:21 21:48:51 scanED001 2890.0 um
2 21:49:03 21:49:33 scanED002 5194.0 um
3 21:49:45 21:50:15 scanED003 8084.0 um
4 21:50:27 21:50:57 scanED004 10314.0 um
5 21:51:09 21:51:39 scanED005 13204.0 um
6 21:51:51 21:52:21 scanED006 13208.5 um
7 21:52:33 21:53:03 scanED007 13214.0 um
8 21:53:15 21:53:45 scanED008 13228.8 um
9 21:53:57 21:54:27 scanED009 13254.5 um
10 21:54:39 21:55:09 scanED010 13303.8 um
11 21:55:21 21:55:51 scanED011 13447.9 um
12 21:56:03 21:56:34 scanED012 13684.5 um
13 21:56:46 21:57:17 scanED013 14180.2 um
14 21:57:29 21:57:59 scanED014 15508.0 um
15 21:58:11 21:58:41 scanED015 15512.5 um
16 21:58:53 21:59:23 scanED016 15518.0 um
17 21:59:35 22:00:05 scanED017 15532.8 um
18 22:00:17 22:00:47 scanED018 15558.5 um
19 22:00:59 22:01:29 scanED019 15607.8 um
20 22:01:41 22:02:11 scanED020 15751.9 um
21 22:02:23 22:02:53 scanED021 15988.5 um
22 22:03:05 22:03:35 scanED022 16484.2 um
23 22:03:47 22:04:17 scanED023 18398.0 um
21/07/01
22:10:00
MT終了
aft21/07/01
1 22:42:26 23:24:22
以上。

p.147
PL 45K, Sor 1K cps

0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:31
0:00:31
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30
0:00:30

12C6+ 135MeV/u 32.6MHz Hyper-ECR 1pnA
p.147

1 / 1 ページvar

SSD@Air2