Page 129 of 228

Report No.: S24060401207001

MEASUREMENT 17
Date of measurement: 24/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Left head
Cheek
LTE band 2
MiddleLTE (Crest factor: 1.0)
2.63

B. SAR Measurement ResultsFrequency (MHz)1880.000000

Relative permittivity (real part)39.096489

Relative permittivity(imaginary part)
Conductivity (S/m)

13.909600

Variation (%)-0.690000

SURFACE SAR

VOLUME SAR

1.452780

Maximum location: X=-6.00, Y=-21.00
SAR Peak: 0.63 W/kg
SAR 10g (W/Kg)
0.240797
SAR 1g (W/Kg)
0.421356

Page 130 of 228

Z (mm)SAR(W/Kg)

0.00
0.63694.00
0.42843D screen shot

9.00
0.255814.00
0.155919.00
0.0937

Report No.: S24060401207001

24.00
0.0587

Hot spot position

29.00
0.0384

Page 131 of 228

Report No.: S24060401207001

MEASUREMENT 18
Date of measurement: 24/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Validation plane
Body
LTE band 2
MiddleLTE (Crest factor: 1.0)
2.63

B. SAR Measurement ResultsFrequency (MHz)1880.000000

Relative permittivity (real part)39.096489

Relative permittivity(imaginary part)
Conductivity (S/m)

13.909600

Variation (%)-1.910000

SURFACE SAR

VOLUME SAR

1.452780

Maximum location: X=7.00, Y=47.00
SAR Peak: 1.10 W/kg
SAR 10g (W/Kg)
0.377855
SAR 1g (W/Kg)
0.693117

Page 132 of 228

Z (mm)SAR(W/Kg)

0.00
1.04114.00
0.6965

3D screen shot

9.00
0.418914.00
0.255119.00
0.1627

Report No.: S24060401207001

24.00
0.0987

Hot spot position

29.00
0.0639

Page 133 of 228

Report No.: S24060401207001

MEASUREMENT 19
Date of measurement: 14/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Left head
Cheek
LTE band 4
MiddleLTE (Crest factor: 1.0)
2.45

B. SAR Measurement ResultsFrequency (MHz)1732.500000

Relative permittivity (real part)

39.692825

Relative permittivity(imaginary part)
Conductivity (S/m)

13.919613

Variation (%)-0.840000

SURFACE SAR

VOLUME SAR

1.339763

Maximum location: X=-6.00, Y=-20.00
SAR Peak: 1.19 W/kg
SAR 10g (W/Kg)
0.462977
SAR 1g (W/Kg)
0.782616

Page 134 of 228

Z (mm)SAR(W/Kg)

0.00
1.18084.00
0.8200

3D screen shot

9.00
0.512014.00
0.314319.00
0.1942

Report No.: S24060401207001

24.00
0.1236

Hot spot position

29.00
0.0754

Page 135 of 228

Report No.: S24060401207001

MEASUREMENT 20
Date of measurement: 14/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Validation plane
Body
LTE band 4
MiddleLTE (Crest factor: 1.0)
2.45

B. SAR Measurement ResultsFrequency (MHz)1732.500000

Relative permittivity (real part)

39.692825

Relative permittivity(imaginary part)
Conductivity (S/m)

13.919613

Variation (%)-2.490000

SURFACE SAR

VOLUME SAR

1.339763

Maximum location: X=10.00, Y=46.00
SAR Peak: 1.47 W/kg
SAR 10g (W/Kg)
0.545415
SAR 1g (W/Kg)
0.947211

Page 136 of 228

Z (mm)SAR(W/Kg)

0.00
1.40444.00
0.9706

3D screen shot

9.00
0.612314.00
0.390519.00
0.2551

Report No.: S24060401207001

24.00
0.1659Hot spot position

29.00
0.1050Page 137 of 228

Report No.: S24060401207001

MEASUREMENT 21
Date of measurement: 21/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Left head
Cheek
LTE band 5
MiddleLTE (Crest factor: 1.0)
2.32

B. SAR Measurement ResultsFrequency (MHz)836.500000

Relative permittivity (real part)

41.672005

Relative permittivity(imaginary part)
Conductivity (S/m)

19.676403

Variation (%)

0.060000SURFACE SAR

VOLUME SAR

0.914406

Maximum location: X=-52.00, Y=-20.00
SAR Peak: 0.10 W/kg
SAR 10g (W/Kg)
0.062710
SAR 1g (W/Kg)
0.083053

Page 138 of 228

Z (mm)SAR(W/Kg)

0.00
0.09814.00
0.08593D screen shot

9.00
0.070314.00
0.056519.00
0.0444

Report No.: S24060401207001

24.00
0.0351

Hot spot position

29.00
0.0283

Page 139 of 228

Report No.: S24060401207001

MEASUREMENT 22
Date of measurement: 21/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Validation plane
Body
LTE band 5
MiddleLTE (Crest factor: 1.0)
2.32

B. SAR Measurement ResultsFrequency (MHz)836.500000

Relative permittivity (real part)

41.672005

Relative permittivity(imaginary part)
Conductivity (S/m)

19.676403

Variation (%)

0.060000SURFACE SAR

VOLUME SAR

0.914406

Maximum location: X=3.00, Y=-55.00
SAR Peak: 0.36 W/kg
SAR 10g (W/Kg)
0.142915
SAR 1g (W/Kg)
0.238887

Page 140 of 228

Z (mm)SAR(W/Kg)

0.00
0.36134.00
0.2517

3D screen shot

9.00
0.160814.00
0.105219.00
0.0716

Report No.: S24060401207001

24.00
0.0496

Hot spot position

29.00
0.0353

Page 141 of 228

Report No.: S24060401207001

MEASUREMENT 23
Date of measurement: 18/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=12mm dy=12mm, h= 5.00 mm
7x7x7,dx=5mm dy=5mm dz=5mm
Left head
Cheek
LTE band 7
MiddleLTE (Crest factor: 1.0)
2.65

B. SAR Measurement ResultsFrequency (MHz)2535.000000

Relative permittivity (real part)

39.361263

Relative permittivity(imaginary part)
Conductivity (S/m)

13.716830

Variation (%)

1.180000SURFACE SAR

VOLUME SAR

1.931787

Maximum location: X=10.00, Y=1.00
SAR Peak: 0.79 W/kg
SAR 10g (W/Kg)
0.179244
SAR 1g (W/Kg)
0.418236

Page 142 of 228

Z (mm)SAR(W/Kg)

0.00
0.90584.00
0.4605

3D screen shot

9.00
0.170314.00
0.109319.00
0.0498

Report No.: S24060401207001

24.00
0.0288Hot spot position

29.00
0.0185

Page 143 of 228

Report No.: S24060401207001

MEASUREMENT 24
Date of measurement: 18/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=12mm dy=12mm, h= 5.00 mm
7x7x7,dx=5mm dy=5mm dz=5mm
Validation plane
Body
LTE band 7
High
LTE (Crest factor: 1.0)
2.65

B. SAR Measurement ResultsFrequency (MHz)2560.000000

Relative permittivity (real part)39.208763

Relative permittivity(imaginary part)
Conductivity (S/m)

13.784530

Variation (%)

4.010000SURFACE SAR

VOLUME SAR

1.960466

Maximum location: X=16.00, Y=57.00
SAR Peak: 2.20 W/kg
SAR 10g (W/Kg)
0.521412
SAR 1g (W/Kg)
1.102988

Page 144 of 228

Z (mm)SAR(W/Kg)

0.00
2.14424.00
1.2877

3D screen shot

9.00
0.644514.00
0.316419.00
0.1620

Report No.: S24060401207001

24.00
0.0808Hot spot position

29.00
0.0448

Page 145 of 228

Report No.: S24060401207001

MEASUREMENT 25
Date of measurement: 20/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Left head
Cheek
LTE band 12
MiddleLTE (Crest factor: 1.0)
2.37

B. SAR Measurement ResultsFrequency (MHz)707.500000

Relative permittivity (real part)41.559696

Relative permittivity(imaginary part)
Conductivity (S/m)

21.927753

Variation (%)-3.110000

SURFACE SAR

VOLUME SAR

0.861883

Maximum location: X=-52.00, Y=-26.00
SAR Peak: 0.08 W/kg
SAR 10g (W/Kg)
0.056534
SAR 1g (W/Kg)
0.070385

Page 146 of 228

Z (mm)SAR(W/Kg)

0.00
0.08084.00
0.0716

3D screen shot

9.00
0.062114.00
0.053619.00
0.0457

Report No.: S24060401207001

24.00
0.0372

Hot spot position

29.00
0.0301

Page 147 of 228

Report No.: S24060401207001

MEASUREMENT 26
Date of measurement: 20/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Validation plane
Body
LTE band 12
MiddleLTE (Crest factor: 1.0)
2.37

B. SAR Measurement ResultsFrequency (MHz)707.500000

Relative permittivity (real part)41.559696

Relative permittivity(imaginary part)
Conductivity (S/m)

21.927753

Variation (%)

0.450000SURFACE SAR

VOLUME SAR

0.861883

Maximum location: X=-10.00, Y=-11.00
SAR Peak: 0.16 W/kg
SAR 10g (W/Kg)
0.091839
SAR 1g (W/Kg)
0.123407

Page 148 of 228

Z (mm)SAR(W/Kg)

0.00
0.14694.00
0.1257

3D screen shot

9.00
0.098014.00
0.070019.00
0.0612

Report No.: S24060401207001

24.00
0.0386

Hot spot position

29.00
0.0389

Page 149 of 228

Report No.: S24060401207001

MEASUREMENT 27
Date of measurement: 20/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Left head
Cheek
LTE band 17
MiddleLTE (Crest factor: 1.0)
2.37

B. SAR Measurement ResultsFrequency (MHz)710.000000

Relative permittivity (real part)

41.544346

Relative permittivity(imaginary part)
Conductivity (S/m)

21.868204

Variation (%)-3.150000

SURFACE SAR

VOLUME SAR

0.862579

Maximum location: X=-52.00, Y=-25.00
SAR Peak: 0.08 W/kg
SAR 10g (W/Kg)
0.055112
SAR 1g (W/Kg)
0.068678

Page 150 of 228

Z (mm)SAR(W/Kg)

0.00
0.07644.00
0.0697

3D screen shot

9.00
0.060814.00
0.052019.00
0.0437

Report No.: S24060401207001

24.00
0.0351

Hot spot position

29.00
0.0290Page 151 of 228

Report No.: S24060401207001

MEASUREMENT 28
Date of measurement: 20/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Validation plane
Body
LTE band 17
MiddleLTE (Crest factor: 1.0)
2.37

B. SAR Measurement ResultsFrequency (MHz)710.000000

Relative permittivity (real part)

41.544346

Relative permittivity(imaginary part)
Conductivity (S/m)

21.868204

Variation (%)

0.330000SURFACE SAR

VOLUME SAR

0.862579

Maximum location: X=-13.00, Y=-10.00
SAR Peak: 0.16 W/kg
SAR 10g (W/Kg)
0.089495
SAR 1g (W/Kg)
0.120952

Page 152 of 228

Z (mm)SAR(W/Kg)

0.00
0.15004.00
0.1226

3D screen shot

9.00
0.095014.00
0.072319.00
0.0552

Report No.: S24060401207001

24.00
0.0437

Hot spot position

29.00
0.0321

Page 153 of 228

Report No.: S24060401207001

MEASUREMENT 29
Date of measurement: 18/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=12mm dy=12mm, h= 5.00 mm
7x7x7,dx=5mm dy=5mm dz=5mm
Left head
Cheek
LTE band 41
MiddleLTE (Crest factor: 1.0)
2.65

B. SAR Measurement ResultsFrequency (MHz)2595.000000

Relative permittivity (real part)

38.993363

Relative permittivity(imaginary part)
Conductivity (S/m)

13.930630

Variation (%)-1.120000

SURFACE SAR

VOLUME SAR

2.008332

Maximum location: X=-24.00, Y=24.00
SAR Peak: 1.05 W/kg
SAR 10g (W/Kg)
0.272640
SAR 1g (W/Kg)
0.575790

Page 154 of 228

Z (mm)SAR(W/Kg)

0.00
1.04304.00
0.61393D screen shot

9.00
0.291214.00
0.139519.00
0.0613

Report No.: S24060401207001

24.00
0.0305

Hot spot position

29.00
0.0188

Page 155 of 228

Report No.: S24060401207001

MEASUREMENT 30
Date of measurement: 18/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=12mm dy=12mm, h= 5.00 mm
7x7x7,dx=5mm dy=5mm dz=5mm
Validation plane
Body
LTE band 41
MiddleLTE (Crest factor: 1.0)
2.65

B. SAR Measurement ResultsFrequency (MHz)2595.000000

Relative permittivity (real part)

38.993363

Relative permittivity(imaginary part)
Conductivity (S/m)

13.930630

Variation (%)-1.650000

SURFACE SAR

VOLUME SAR

2.008332

Maximum location: X=-26.00, Y=44.00
SAR Peak: 0.68 W/kg
SAR 10g (W/Kg)
0.195241
SAR 1g (W/Kg)
0.386248

Page 156 of 228

Z (mm)SAR(W/Kg)

0.00
0.68034.00
0.40383D screen shot

9.00
0.201914.00
0.100619.00
0.0577

Report No.: S24060401207001

24.00
0.0263

Hot spot position

29.00
0.0170Page 157 of 228

Report No.: S24060401207001

MEASUREMENT 31
Date of measurement: 14/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Left head
Cheek
FDDBand66
Middle(Crest factor: 1.0)
2.45

B. SAR Measurement ResultsFrequency (MHz)1745.000000

Relative permittivity (real part)

39.621025

Relative permittivity(imaginary part)
Conductivity (S/m)

13.917163

Variation (%)-0.500000

SURFACE SAR

VOLUME SAR

1.349192

Maximum location: X=-6.00, Y=-20.00
SAR Peak: 1.14 W/kg
SAR 10g (W/Kg)
0.435590
SAR 1g (W/Kg)
0.721024

Page 158 of 228

Z (mm)SAR(W/Kg)

0.00
1.12334.00
0.78483D screen shot

9.00
0.487714.00
0.297819.00
0.1795

Report No.: S24060401207001

24.00
0.1152

Hot spot position

29.00
0.0733

Page 159 of 228

Report No.: S24060401207001

MEASUREMENT 32
Date of measurement: 14/6/2024

A. Experimental conditions.
Area Scan
ZoomScanPhantom
Device Position
Band
Channels
SignalConvF

dx=15mm dy=15mm, h= 5.00 mm
5x5x7,dx=8mm dy=8mm dz=5mm
Validation plane
Body
FDDBand66
Low(Crest factor: 1.0)
2.45

B. SAR Measurement ResultsFrequency (MHz)1720.000000

Relative permittivity (real part)39.815224

Relative permittivity(imaginary part)
Conductivity (S/m)

13.910263

Variation (%)-0.550000

SURFACE SAR

VOLUME SAR

1.329203

Maximum location: X=14.00, Y=48.00
SAR Peak: 1.60 W/kg
SAR 10g (W/Kg)
0.582347
SAR 1g (W/Kg)
1.015335

Page 160 of 228

Z (mm)SAR(W/Kg)

0.00
1.71604.00
1.03283D screen shot

9.00
0.549814.00
0.400519.00
0.2513

Report No.: S24060401207001

24.00
0.1636

Hot spot position

29.00
0.1123

Page 161 of 228

14.

Appendix D. Calibration Certificate
Table of contents

E Field Probe - 3423-EPGO-426
750 MHz Dipole - SN 03/15 DIP 0G750-355
835 MHz Dipole - SN 03/15 DIP 0G835-347
1800 MHz Dipole - SN 03/15 DIP 1G800-349
1900 MHz Dipole - SN 03/15 DIP 1G900-350
2450 MHz Dipole - SN 03/15 DIP 2G450-352
2600 MHz Dipole - SN 03/15 DIP 2G600-356
5000-6000 MHz Dipole - SN 13/14 WGA 33

Report No.: S24060401207001

Page 162 of 228

Report No.: S24060401207001

Page 163 of 228

Report No.: S24060401207001

Page 164 of 228

Report No.: S24060401207001

Page 165 of 228

Report No.: S24060401207001

Page 166 of 228

Report No.: S24060401207001

Page 167 of 228

Report No.: S24060401207001

Page 168 of 228

Report No.: S24060401207001

Page 169 of 228

Report No.: S24060401207001

Page 170 of 228

Report No.: S24060401207001

Page 171 of 228

Report No.: S24060401207001

Page 172 of 228

Report No.: S24060401207001

Page 173 of 228

Report No.: S24060401207001

Page 174 of 228

Report No.: S24060401207001

Page 175 of 228

Report No.: S24060401207001

Page 176 of 228

Report No.: S24060401207001

Page 177 of 228

Report No.: S24060401207001

Page 178 of 228

Report No.: S24060401207001