Official Copy

Approved for Use

Effective Date: 21 Dec 2018

Revision: A

Technical Bulletin
Changes Made to the Triage MeterPro Manual for Software Version 5.04.021

1. New features in the software update include: Sample Select Restore Factory Defaults
2. Update for International Quidel AssistanceSample SelectIn certain cases, the lab supervisor may wish to control the default sample type for devices run on themeter. Sample Select allows the supervisor to choose BLOOD, PLASMA, or SELECT options. If BLOOD orPLASMA are selected, then any device run on the meter for PATIENT SAMPLE or MISC will use the chosen
sample type option without prompting the user to make a selection with each test.
If SELECT is chosen, then each time a user runs a PATIENT SAMPLE or MISC test the user will be prompted
to select BLOOD or PLASMA prior to the test being executed by the meter.
This feature is available on software version 05.04.021 however this feature is not available for all tests.
If this feature is not available for a test, no selection may be made.
1. From the list of programmable parameters, choose SAMPLE TYPE using the2. Press thekey.

3. Select the test panel type using thekeys.

4. Navigate down to the SAMPLE TYPE parameter using thekeys

5. Select the SAMPLE TYPE option of BLOOD, PLASMA, or SELECT using the6. Press thekeys.

keys.

key to save changes.

NOTE: The SAMPLE TYPE that is selected by the supervisor will appear on the result screen andwill also appear in the printed results at the end of the test being performed.

Page 1 of 2

Printed on: 26 Mar 2019, 12:21:27 pm

Official Copy

Approved for Use

Effective Date: 21 Dec 2018

Revision: ARestore Factory Defaults
This function allows the supervisor to restore the meter to factory default settings. All results and data
will be deleted if this setting is selected. All settings will be reset to factory defaults. This option should
only be selected when the Su pervisor desires to delete all data and results stored on the meter. Thisfeature is available on software version 05.04.021.
1. From the list of programmable parameters, choose RESTORE FACTORY DEFAULTS using thekeys.
2. Press thekey.

3. Select RESTORE FACTORY DEFAULTS using the
4. Press thekeys.

key.

NOTE: If the RESTORE FACTORY DEFAULTS selection is chosen and the enter key pressed, allresults and data currently stored on the meter will be deleted permanently.
AssistanceIf you have any questions regarding the use of this product, please contact Quidel Technical Support at+353 (91) 412 474 (outside the U.S.) or one of the Technical Support Centers listed below. You may also
contact us at quidel.com or emeatechnicalsupport@quidel.com.
RegionTelephone Number

Fax Number(Option 2 for Technical Support)
Austria+43 316 231239+43 316 231240

France0 (805) 371674+33 (1) 856 40308

Germany+49 7154 1593912+49 7154 1593913

Ireland (Main)+353 (91) 412 474+353 (91) 412 478

Ireland (Toll-Free)

1800 200441+353 (91) 412 478

Netherlands

0 800 0224198+31 20 888 1320

Switzerland

0 800 554864+41 31 560 2913

United Kingdom

0 800 3688248+44 (20) 397 04114

Europe, Middle Eastand Africa+353 (91) 412 474+353 (91) 412 478

E-mail: emeatechnicalsupport@quidel.com
You may also visit our website at quidel.com for information on Quidel’s line of Rapid Diagnostics ,Molecular Diagnostics, Cell Culture and Specialty Products (Bone Health and Autoimmune &Complement). Other product information available on our website includes: CPT codes, CLSI procedure
guides, SDS, and Package Inserts.

Page 2 of 2

Printed on: 26 Mar 2019, 12:21:27 pm

ART-26641 Rev. A (12/18)

Official Copy

Approved for Use

Effective Date: 21 Dec 2018

Revision: A

Signature Manifest
Document Number: TechBulletin-26641enTitle: Quidel Triage MeterProTech Bulletin for Software Version 5.04.021

Revision: AAll dates and times are in US/Pacific.

ECO-6015 T2 CustomerLetter-26638en TechBulletin-26641en CR-8013

Collaboration
Name/Signature

Title

Alyna Fainberg (AFAINBERG)

Date

Meaning/Reason

06 Dec 2018, 06:29:57 PM

Complete

Justin Stelzer (STELZJU)

Project Manager

07 Dec 2018, 08:23:04 AM

Complete

Alexander Taam
(ALEXANDER.TAAM)ACT Program Lead

07 Dec 2018, 08:44:56 AM

Complete

Payal Aggarwal-Scott (PAYALAS) Quality Assurance Engineer V

07 Dec 2018, 10:13:32 AM

Complete

Jacqueline Paulo-Kretzer(JACQUELINE.KRETZER)

07 Dec 2018, 10:20:14 AM

Complete

Laura DuPont (LAURA.DUPONT) Training Systems Administrator

07 Dec 2018, 11:41:09 AM

CompleteDawn Allenby (ALLENDA)

Associate Director,Regulatory

07 Dec 2018, 02:11:15 PM

Complete

Billy Zarcone (BILLY.ZARCONE)

Access Only

07 Dec 2018, 03:18:06 PMComplete

07 Dec 2018, 03:54:17 PM

Complete

Document Control Analyst

07 Dec 2018, 04:20:17 PM

Complete

Title

Date

Meaning/Reason

Technical Marketing Manager

Jody Vacala (JODY.VACALA)
Deborah Phillips (PHILLDE)

Approvals
Name/Signature

Payal Aggarwal-Scott (PAYALAS) Quality Assurance Engineer V

07 Dec 2018, 04:30:55 PM

Approved

Jody Vacala (JODY.VACALA)

07 Dec 2018, 04:50:07 PMApproved

Alexander Taam(ALEXANDER.TAAM)ACT Program Lead

08 Dec 2018, 11:05:00 AM

Approved

Justin Stelzer (STELZJU)

Project Manager

10 Dec 2018, 09:37:38 AM

Approved

Jacqueline Paulo-Kretzer(JACQUELINE.KRETZER)

Technical Marketing Manager

11 Dec 2018, 02:59:51 AM

Approved

Dawn Allenby (ALLENDA)

Associate Director,Regulatory

11 Dec 2018, 01:55:39 PM

Approved

Alyna Fainberg (AFAINBERG)

14 Dec 2018, 02:59:41 AM

Approved

Nenita Trinidad (TRININE)

Manager, Manufacturing I

21 Dec 2018, 05:16:22 PM

Approved

Deborah Phillips (PHILLDE)

Document Control Analyst

21 Dec 2018, 06:57:10 PM

Approved

Title

Date

Meaning/Reason

DMR
Name/Signature
Erwin Gobaleza(ERWIN.GOBALEZA)

Document Control Analyst

Deborah Phillips (PHILLDE)

Document Control Analyst

21 Dec 2018, 07:01:00 PM

Approved

Title

Date

Meaning/Reason

Document Control Analyst

21 Dec 2018, 07:01:18 PM

Approved

Final Release
Name/Signature
Deborah Phillips (PHILLDE)

Printed on: 26 Mar 2019, 12:21:27 pm