Universal High Performance Guard Cartridge Column for HPLCSunGuard

Can be used regardless ofcolumn manufacturer

seriesSunGuard Prep for 10 mm to 20 mm I.D. preparative columns

NEW*Simple structure*Low dead volume*The plate number is maintained even after the guard
cartridge column is connected.
*Can be used with 10 mm and 20 mm inner diameter
preparative columns*Upper Pressure limit is 30 MPa.
*Easy installation*Set a starter kit with a 21mm spanner*Select one coupler tubing in the starter kit from 50mm, 100 mm, and 150 mm lengths.

SpannerHex 21 mmHolderCartridge
10 mm I.D.
10 mm length

Coupler
0.8 mm I.D.
50 mm length

SUS tubing for coupler
50 mm，100 mm，150ｍｍ length

SunGuard for 4.6 mm I.D. analytical columns*Simple structure*Low dead volume*The plate number is maintained even after theguard cartridge column is connected.
* Can be used with 3 μm columns as well as 5 μmcolumns.
*Upper Pressure limit is 40 MPa.
*Easy installation (put the cartridge in the holder,
1/4 turn with a wrench after tightening by hand)*SunArmor type has high alkali durabilityspecification.

Hex 14 mm
PEEK O-ring

HolderFilter

カートリッジ(4x10 mm)の概略図

Cartridge
4.0 mm I.D.
10 mm length

SunGuard Ultra for high-throughput columns or small-diameter columns
HolderHex 12 mm

Cartridge
2.0 mm I.D.
3 mm length

Coupler
0.13 mm I.D.
50 mm length*“RP” (reversed phase) is SunShell C18, and “S” is SunShell HILIC-S coreshell silica packed in a cartridge with an inner diameter of 2 mm and alength of 3 mm.
*”RP” can be used for all reversed-phase columns, and “S” can be usedfor all hillic columns.
*Low dead volume* Upper Pressure limit is 60 MPa or more.
*Can be used for columns with an inner diameter of 2.1 mm to 4.6 mm※ Use SunGuard Ultra for core-shell columns (SunShell) and high-speed analysis columns
with small retention volumes.

*Ordering information is displayed on the back.

Chromanik Technologies Inc.

SunGuard

Ordering information

SunGuard Prep for 10 mm to 20 mm I.D. preparative columns
Particle
size

Catalog number

Sunniest, SunArmor, Sunrise prep guard cartridge column starter kit (without spanner)(holder, cartridge 10 x 10 mm, coupler [length 50mm] each one)

5 µm⃞⃞371K1

Sunniest, SunArmor, Sunrise prep guard cartridge column starter kit (without spanner)(holder, cartridge 10 x 10 mm, coupler [length 100mm] each one)

5 µm⃞⃞371K2

Sunniest, SunArmor, Sunrise prep guard cartridge column starter kit (without spanner)(holder, cartridge 10 x 10 mm, coupler [length 150mm] each one)

5 µm⃞⃞371K3

Sunniest, SunArmor, Sunrise prep guard cartridge column starter kit (with spanner)(holder, cartridge 10 x 10 mm, coupler [length 50mm] each one. and two spanners)

5 µm⃞⃞371K1S

Sunniest, SunArmor, Sunrise prep guard cartridge column starter kit (with spanner)(holder, cartridge 10 x 10 mm, coupler [length 100mm] each one. and two spanners)

5 µm⃞⃞371K2S

Sunniest, SunArmor, Sunrise prep guard cartridge column starter kit (with spanner)(holder, cartridge 10 x 10 mm, coupler [length 150mm] each one. and two spanners)

5 µm⃞⃞371K3S

Sunniest, SunArmor, Sunrise guard cartridge (2 pieces) 10 x 10 mm

5 µm⃞⃞371CSunGuard Prep holder 10 x 10 mm---HOL71C

DescrptionBlanks (⃞⃞ ) in the model number contain the 2-digit code corresponding to packings below.
Sunniest C18

EB

Sunniest PhE (Phenyl)EP

Sunniest Biphenyl

E8

SunArmor C18

HB

Sunniest C8

EC

Sunniest PFPEFSunniest Silica

ES

SunArmor RP-AQUA

HR

Sunniest RP-AQUA

ER

Sunniest PFP&C18

EV

Sunrise C30 (C28)SM

SunArmor NH2

HN

SunGuard for 4.6 mm I.D. analytical columns
Particle
size

CatalognumberSunniest, Sunrise, SunArmor guard cartridge column (Holder, cartridge, each one) 4 x 10 mm

5 µm⃞⃞3A1H

Sunniest, Sunrise, SunArmor cartridge for exchange (4 pieces) 4 x 10 mm

5 µm⃞⃞3A1CSunGuard guard cartridge column holder 4 x 10 mm---

HOLA1C

DescrptionBlanks (⃞⃞ ) in the model number contain the 2-digit code corresponding to packings below.
Sunniest C18

EB

Sunniest PhE (Phenyl)EP

Sunniest Biphenyl

E8

SunArmor C18

HB

Sunniest C8

EC

Sunniest PFPEFSunniest Silica

ES

SunArmor RP-AQUA

HR

Sunniest RP-AQUA

ER

Sunniest PFP&C18

EV

Sunrise C30 (C28)SM

SunArmor NH2HN

SunGuard Ultra for high-throughput columns or small-diameter columns
CatalognumberDescrption

SunShell Guard cartridge RP Starter Kit (Holder, Cartridge 2 x 3 mm, Coupler, each one）

CB32CKSunShell RP cartridge for exchange (2 pieces) 2 x 3 mm

CB32CCSunShell Guard cartridge S Starter Kit (Holder, Cartridge 2 x 3 mm, Coupler, each one）

CS32CK

SunShell S cartridge for exchange (2 pieces) 2 x 3 mm

CS32CCSunGuard Ultra holder 2 x 3 mm

HOL2CC*The holder is not included with the “guard cartridge for exchange". A holder is required for use.
*For SunGuard Prep/SunGuard Ultra, we recommend the starter kit, which comes with a complete set of connection
parts, for the first time. SunGuard does not have a starter kit.
*Please note that the appearance and specifications are subject to change without notice.

Manufacturer

ChromaNik Technologies Inc.

2405

6-3-1 Namiyoke, Minato-ku, Osaka, 552-0001 Japan
TEL: +81-6-6581-0885 FAX: +81-6-6581-0890
E-mail: info@chromanik.co.jp
URL: http://chromanik.co.jp