[3] NAME OF COMPONENT UNITS
3-1 Main unit
Clamping current sensor(CT)
Center position indication

DCL11R

300V CAT.III MAX 300A

Barrier
HOLD button

Trigger
Power button

DIGITAL CLAMP METER

Backlight button

INSTRUCTION MANUAL
LCD

3-2 Display

This manual emplys soy ink.

Data HOLD

Auto power OFF enabled

Dempa Bldg., 4-4 Sotokanda 2-Chome
Chiyoda-ku, Tokyo, Japan

Low battery indication

AC (Alternate current)

Unit (ampere)

01-1507 2040 6011

[1] SAFETY PRECAUTIONSBefore use, read the following safety precautions.
This instruction manual explains how to safely use your new DCL11R digital clamp
meter. Before use, please read this manual thoroughly. After reading it, keep it
together with the product so you can refer to it when necessary. If this product is not
used as specified in this manual, its protection function may be compromised.
To avoid accidental burns or electric shock, always follow any instructions with
“ WARNING” or “ CAUTION” headings.
1-1 Explanation of Warning Symbols
The meanings of the symbols used in this manual and on the product are explained below.
: Very important instructions for safe use•Warning messages are intended to prevent accidents to operating
personnel such as burn and electric shock.
•Caution messages are intended to prevent damage to the instrument.
The meanings of the symbols used exclusively on the product are explained below.
: Symbol requesting reference to the instruction manual before use: Potential of connection to a power line in live status: Double in sulation or reinforced insulation: Power switch: Backlight
1-2 Warning Instructions for Safe Use
WARNINGTo avoid physical injury such as burns or electric shock, be sure to observe
the following instructions when using this instrument.
1. This instrument is a digital clamp meter for low voltages. It should be
used on power lines with a voltage to ground of 300 Vrms or less.
2. Pay special attention when measuring voltages of AC 33 Vrms (46.7 V peak)or DC 70 V or more to avoid injury. Also use insulating protective equipment
as required.
3. Never use the instrument on power lines that exceed the maximum rated
input current (see 1-3).
4. Do not use the instrument if the main unit is damaged or broken.
5. Do not use the instrument with the case or battery compartment cover removed.
6. Do not hold the measurement clip at any point beyond the barrier.
7. Do not use the instrument when it or your hand is wet.
8. Do not attempt to repair or modify the instrument except to replace the batteries.
9. Be sure to check the instrument before each use and inspect it at least once a year.
10. Always use the instrument indoors.
11. To avoid compromising the protection function of this instrument, do notuse it in any way other than instructed in this manual.
* Voltage to ground of 300 Vrms or less: When the Y-connection voltage is 415 V,
the voltage to ground is equal to 415 / 3 ≒ 240 V.

1-3 Overload Protection Input Values
Input

Max. rated input current

Max. overload protection input

Clamp sensor

AC 320 A

AC 350 A[2] APPLICATIONS AND FEATURES

Numeric value & decimal point[4] DESCRIPTION OF FUNCTIONS
4-1 Power button:
buttonPress the button to switch the power on and off.
4-2 Auto power OFF function
The instrument incorporates the auto power OFF function that turns the
instrument OFF in about 15 minutes after the last operation. This function canbutton. The
be disabled by switching the instrument ON while holding theindicator on the display panel is extinguished when the auto power OFF
function is disabled.
4-3 Low battery indication
The
indicator lights on the display when the batteries have discharged
down to a supply voltage of about 2.3 V or less. Replace the batteries when
this indicator lights up.
4-4 Data hold function:
buttonPress the HOLD button. The
indicator lights on the display and the value
displayed at that time is held and will not change even if the measurement
indicator anddata varies. Pressing the button against extinguishes thethe measurement data is no longer held.
4-5 Backlight function:
buttonPress this button to light the backlight of the display. Pressing the button again
extinguishes the backlight. The backlight is also extinguished automatically inabout 10 seconds after it is lit.

[5] MEASURING PROCEDUREWARNING
1. Do not apply an input voltage exceeding the maximum rated current.
2. During measurement, do not hold the measurement clip at any point beyond the barrier.

5-1 Start-up inspection
Perform the following pre-operational check for safety.
• Check the external appearance. Check that the main unit is free of abnormality
or damage due to dropping, etc.
indicator is not lit. If it is lit, replace the batteries with new ones.
• Check that the• Make sure your hands and the instrument are not wet.
*If nothing is displayed on the LCD, the batteries might have been totally discharged.
5-2 Measuring the AC current (ACA measurement)
CAUTION
1. To improve the measuring accuracy, position the conductor at the center ofthe clamp sensor.
2. Always clamp only one wire at a time. Accurate measurement is impossible if two
or more wires, a cable with multiple conductors or a parallel cord is clamped.
3. Accurate measurement is impossible if the clamp sensor is not closed completely.
4. Accurate measurement is sometimes impossible near a source of strong
magnetic field such as a transformer or high-current line, near a source ofelectromagnetic waves such as radio equipment or near a charged object.
5. The CT may generate an oscillating sound when a high current is applied.
This is not malfunction.

2-1 Application
This instrument is an AC clamp meter with RMS response that allows you tomeasure the current range specified in IEC 61010-1 CAT. III 300 V. It is suitable
for measuring the current of low-voltage power lines and electrical appliances.
2-2 Features• Safety design that complies with IEC61010-1 CAT. III 300 V• Compact pocket size
• True RMS conversion• Auto power OFF (approx. 15 min.) (Disabling also possible)• Backlight
Measurement Category (Overvoltage Category)Equipment of CAT II :Primary cable runs of power-consuming equipment from a wall socket.
Equipment of CAT III :Primary cable runs of equipment directly connected to a distribution board
and cable runs from a distribution board to wall sockets.
Equipment of CAT IV : Cable runs from an incoming line to a distribution board.

① Set the Power switch to ON.
② Open the clamp sensor, catch the single wire to be measured and close theclamp sensor completely.
③ Read the measurement value.

− 1 −

− 2 −

Incorrect ×

Clamping single conductor on
the center of clamp sensor.

Load

Load

Correct ○Clamping multiple conductors.

Force applied to the tip ofclamp sensor.

[6] MAINTENANCEWARNING
1. This section is very important for safety. Read and understand the following
instructions fully and maintain your instrument properly.
2. To maintain safety and accuracy, calibrate and inspect the instrument atleast every year.

6-1 Maintenance and Inspection
Before use, check the instrument to confirm that there is no abnormality and that
the CT can be opened and closed smoothly. If anything is abnormal, do not use theinstrument and return it to your authorized Sanwa agent or distributor for repair.
6-2 Calibration
For more information, please contact Sanwa’s authorized agent / distributeservice provider, listed in our website. See section 7-3.
6-3 Cleaning and Storage
CAUTION
1. The panel and case are not resistant to volatile solvent and must not becleaned with thinner or alcohol. If the instrument gets dirty, wipe with a soft
cloth moistened with a small amount of water.
2. The panel and case are not resistant to heat. Do not place the instrument
near heat-generating devices (such as a soldering iron).
3. Do not store the instrument anywhere it may be subject to vibrations or could fall.
4. When storing the instrument, avoid hot, cold or humid locations, locations
exposed to direct sunlight, or locations where condensation is anticipated.
5. When the instrument is not going to be used for an extended time, be sure
to remove the batteries.

• The service charge or transport freight could sometimes become higher than the
product price. Please consult us before asking for servicing.
• The minimum retention period of the servicing performance parts of thisproduct is six (6) years after the discontinuation of production. This period isequal to the servicing available period. However, the retention period of a partmay be reduced if it becomes unavailable due to discontinuation of productionof the part manufacturer, etc.
4) Precautions when sending the product to be repaired:
To ensure the safety of the product during transportation, place the product ina box that is larger than the product 5 times or more in volume and fill cushion
materials fully and then clearly mark “Repair Product Enclosed” on the box
surface. The cost of sending and returning the product shall be borne by thecustomer.
7-3 SANWA web site
http://www.sanwa-meter.co.jp
E-mail: exp_sales@sanwa-meter.co.jp[8] SPECIFICATIONS
8-1 General Specifications
Operation method

Δ -Σ method

AC detection method

True RMS method (AC coupling)

Display

6000 countSampling rate

Approx. 2 times/sec

Overload display“OL” on the numeric display

Range selection

Auto range

Low battery indication

When the batteries are discharged down to 2.3 V orless, theindicator lights on the display.

Current measurement
method / aperture
diameterClamping current sensor (CT) /Max. approx. dia. φ 22 mm

Operation
environment

Altitude max. 2000 m, indoor use, environmental
pollution degree II.

Operatingtemperature/humidity

Temperature 5 to 40 °C,
Relative humidity max. 80 % RH at 5-31 °C anddecreases linearly through 80 % RH to 50 % RH at
31-40 °C (without condensation)

Storage temperature/humidity-10 to 40 °C: Max. 80 % RH (without condensation)
40 to 50 °C: Max. 70 % RH (without condensation)

Power supply“AAA”-size alkaline battery LR03 x 2[7] AFTER-SALE SERVICE

Auto power OFF

Power off in approx. 15 min. after last operation

7-1 Warranty and provision
Sanwa offers comprehensive warranty services to its end-users and to its
product resellers. Under Sanwa's general warranty policy, each instrument iswarranted to be free from defects in workmanship or material under normal use
for the period of one (1) year from the date of purchase.
This warranty policy is valid within the country of purchase only, and applied
only to the product purchased from Sanwa authorized agent or distributor.
Sanwa reserves the right to inspect all warranty claims to determine the extent
to which the warranty policy shall apply. This warranty shall not apply todisposables batteries, or any product or parts, which have been subject to oneof the following causes:
1. A failure due to improper handling or use that deviates from the instruction manual.
2. A failure due to inadequate repair or modification by people other than Sanwa
service personnel.
3. A failure due to causes not attributable to this product such as fire, flood andother natural disaster.
4. Non-operation due to a discharged battery.
5. A failure or damage due to transportation, relocation or dropping after thepurchase.

Power consumption /Battery life

Approx. 1.9 mA / Approx. 500 hours (backlight
extinguished)

Dimensions and mass

145 (H) x 54 (W) x 31 (D) mm, approx. 120 g(including batteries)

Safety standards

IEC61010-1, IEC61010-2-030 CAT. III 300 V,
IEC61010-2-32

EMC directive

IEC61326-1

Accessories

Instruction manual, carrying pouch (C-DCL10)

6-4 Battery replacement• Batteries shipped with the instrument
The batteries loaded at the factory are monitor batteries, so their service life
may be shorter than that of brand-new batteries. Monitor batteries are a typeof battery used to check the functions and performance of a product.
WARNING
1. To avoid electric shock, do not remove the battery compartment cover while an
input is applied to the clamp sensor or the instrument is measuring something.
2. Make sure that the instrument is set to OFF before replacing the batteries.
① Using a Phillips screwdriver, remove thef i x i n g s c re w ( x 1 ) f ro m t h e b a t t e r ycompartment cover.
② Remove the battery compartment cover.
③ Replace both of the batteries in thebattery holder with new ones byobserving the correct polarity.
④ Place the battery compartment coverand tighten the fixing screw in theoriginal position.

7-2 Repair
Customers are asked to provide the following information when requesting
services:
1. Customer name, address, and contact information
2. Description of the problem
3. Description of the product configuration
4. Model Number
5. Product Serial Number
6. Proof of Date-of-Purchase
7. Where you purchased the productPlease contact Sanwa authorized agent / distributor / service provider, listed
in our website, in your country with above information. An instrument sent to
Sanwa / agent / distributor without above information will be returned to thecustomer.
Note:1) Prior to requesting repair, please check the following:
Capacity of the built-in battery, polarity of installation and discontinuity of thetest leads.
2) Repair during the warranty period:
The failed instrument will be repaired in accordance with the conditions
stipulated in 7-1 Warranty and provision.
3) Repair after the warranty period has expired:• If it is expected that servicing can restore the original functioning of the product,
we will service it for a price upon request of the user.
− 3 −8-2 Measuring ranges and accuracies
Accuracy-guaranteed temperature/humidity range: 23 ±5 °C, max. 80 % RH
(without condensation).
rdg: Reading
dgt: digits (lowest digit)
Accuracy-guaranteed range: 1 % to 100 % of measurement range
Crest factor CF: Full-scale CF < 1.6, half scale CF < 3.2

ACA measurement RMS response, AC coupling (sine wave alternate current)
Range
60.00 A
300.0 A

Resolution
0.01 A
0.1 A

Accuracy: 50/60 Hz

Accuracy: 45-400 Hz±(2.0 %rdg + 5 dgt)±(3.0 %rdg + 10 dgt)

Accuracy measurement method(Example) ACA measurement
Reading: 10.00 A
Range accuracy: 60.00 A range = ±(2 %rdg + 5 dgt)
Error: ±(10.00 A x 2.0 % + 0.01 A + 5) = ±0.25 A
True value: 10.00 A ±0.25 A (in the range from 9.75 to 10.25 A)* 3 dgt in the 60.00 A range is equivalent to 0.03 A.
* 3 dgt in the 300.00 A range is equivalent to 0.3 A.

Specifications and external appearance of the product described above
may be revised or modified without prior notice.

− 4 −