PACE 1001 barometer

Precision barometric indicator and recorder
A high precision Druck indicator and recorder,
designed for barometric monitoring inlaboratories, airfields and other applications• Airfield task as standard: Display QFE, QFF or QNH inpressure units or as altitude in feet or meters

Features• Switch test/analogue output option• Utilizes Druck's unique range of resonant pressure sensor
technology• Pressure ranges 750-1150 mbar (10.9-16.7 psi, 75-115 Kpa),
absolute• Choice of precision up to 0.020 mbar or 0.0002901 psi• Long term stability of 0.05 mBara/0.00072515 psi per annum• Selectable numeric or graphic display• High resolution touch screen operation• Intuitive icon driven task menu structure• Leak test option• Switch test/voltage free contact option• RS232, IEEE connectivity, Ethernet and USB as standard• Min/max/average display• Compatible with software packages• 28 selectable pressure units plus 4 user defined units• Various service support options available

PACE 1001 precision barometricindicator and recorder

The PACE 1001 precision barometric indicator brings together
the latest measurement technology from Druck to offer anelegant, fast, flexible and economical solution to barometricindication and monitoring.

Switch test – standard with the analogue output
or voltage free contacts option
Switch test automates the testing of pressure switch devices.
Following the test, displayed is the pressure at whichcontacts open and close and the switch hysteresis. Switch
test task can also be set to capture switch toggle maximum,minimum and average values.

PACE 1001 barometer employs digitally characterizedpressure sensors which offer the quality, stability andprecision associated with this latest generation of resonant
devices.
The PACE 1001 barometer offers three levels of precision toaccommodate specification and budget requirements,
optional external IDOS universal pressure modules can beconnected to add more pressure measurement capability.
The color touch screen display of the PACE 1001 barometercan be configured by a user to indicate pressure
measurement either graphically or numerically.
The measurements displayed can be logged to an internal
memory over a customized time period, sample rate andtrigger ready for replay on the display, can also be saved to aUSB storage device or exported to a PC.
Pressure sensor measurements can be re-transmitted via
one of the communication ports.
The Airfield task is supplied as standard with the PACE 1001
barometer and enables the user to display QFE, QFF or QNH inpressure units or as altitude in feet or meters.

PACE 1001 barometer options
Leak test
Leak test measures leak rate over the measure dwell time.
At the start of the test, the instrument measures the applied
test pressure of the user system. The instrument records thepressure change during measure dwell time. On completionthe display shows the leak rate results, with leak rate persecond or per minute in the current pressure units selected.

Analogue output
The analogue output can be programmed via the setup
menu screen to output a signal proportional to theinstrument range selected. This allows the instrument tointerface with PC or PLC I/O cards, remote displays, chart
recorders or other data logging equipment.
Users can select outputs of 0 to 10 V, 0 to 5 V, -5 to 5 V and
0/4 to 20 mA. Precision with respect to host instrument
measured pressure 0.05% FS over the host instrument
operating temperature range, variable update rate to 80
readings per second. The option is programmable between
minimum and FS pressure for proportional output againstpressure.

Volt free contacts
Volt free contacts enable control of peripheral devices suchas vacuum pumps, ovens, etc. Each VFC option has three
independent volt free NO/NC relay contacts. A number ofconditions can be set within a PACE 1001 instrument to triggera relay toggling its contacts.

Specifications
Pressure measurement
Barometer pressure range

750-1150 mbar absolute, 10.9-16.7 psi absolute, 75-115 kPa absolute

Over range indication:

10% above mbar/bar full scale pressure range

Pressure media:

Dry, oil free, non-corrosive gas, airDisplay
Panel¼ VGA wide format 4.3 inch color graphics LCD c/w integral touch screen

Comms update rate

8 times per second

Display update rate readout

2 times per second± 9999999

Pressure units

mbar, bar, Pa(N/m2), hPa, kPa, MPa, mm Hg @ 0°C, cm Hg @ 0°C, m Hg @ 0°C, in Hg @ 0°C, mm H2O @ 4°C, cm H2O @ 4°C, m H2O@ 4°C, mm H2O @ 20°C, cm H2O @20°C, m H2O @ 20°C, kg/m2, kg/cm2, torr, atm, psi, lb/ft2, in H2O @ 4°C, in H2O @ 20°C, in H2O @60°F, ft H2O @ 4°C, ft H2O @ 20°C, ft H2O @ 60°F, user defined 1, user defined 2, user defined 3, user defined 4 (feet and meters inairfield task)

Performance over the calibrated temperature range
PACE 1001 barometer standard
precisionStandard precision is 0.10 mbar or 0.001450 psi. Includes non-linearity, hysteresis, repeatability and temperature effects between
15°C (59°F) and 45°C (113°F)

PACE 1001 barometer highprecisionHigh precision is 0.05 mbar or 0.000725 psi. Includes non-linearity, hysteresis, repeatability and temperature effects between 15°C
(59°F) and 45°C (113°F)

PACE 1001 barometer premium
precision

Premium precision is 0.025 mbar or 0.0003625 psi. Includes non-linearity, hysteresis, repeatability and temperature effects
between 15°C (59°F) and 45°C (113°F)

PACE 1001 barometer reference
precision

Reference precision is 0.020 mbar or 0.0002901 psi. Includes non-linearity, hysteresis, repeatability and temperature effects
between 15°C (59°F) and 45°C (113°F)

PACE 1001 barometer long termstability

0.05 mBar / 0.00072515 psi per annum

PACE 1001 barometer reference
accuracy

Accuracy (2 Sigma) over calibrated temperature range 0.06 mbar / 0.00087022 per annum.
Includes measurement precision, measurement long term stability (see above) and calibration equipment expanded uncertainty.

ElectricalPower supply

90 VAC to 130 VAC @ 47 to 63 Hz and 180 VAC to 260 VAC @ 47 to 63 Hz. 15 VA

Communications
Communication

RS232, USB and IEEE-488, SCPI, DPI141, DPI142 and DPI150 emulation. LabVIEW drivers ethernet (VXI-II and sockets using SCPI)

Data log
Data log

Display screen shot stored in CSV format, onto memory card or external USB storage device. User defined update rate from 1 second

Environmental
Temperature

Operating 		

10°C to 50°C (50°F to 122°F)

Calibrated

15°C to 45°C (59°F to 113°F)

Storage		-20°C to 70°C (-4°F to 158°F)

Sealing

IP20 (EN60529), indoor use only

Humidity

5% RH to 95% RH non-condensing

Vibration

Compliant with def. stan. 66-31 8.4 Cat 3 and MIL-T-28800E cat 2

Shock

Mechanical shock conforms to EN61010

Conformity

Electrical safety - Global (IEC61010-1, UL61010-1, CSA 22.2, No. 61010-1 and CB test certificate), LVD (EN 61010-1). EMC EN61326, PED,
ROHS and WEEE. CE marked

PhysicalWeight

3.2kg (excluding external PSU and packaging) to 6.5 lbs (including external PSU and packaging)

Dimensions

218 mm wide x 88mm (2U) high x 250 mm deep (8.6in x 3.5 (2U) x 9.8 in)

Pressure connection

G 1/8 Female (1/8 NPT female by adapter, standard for North America)Ordering information

Please state the following (where applicable)

1. Model PACE 1001 barometer
One internal barometric sensor• I1001STANDARD-BARO—standard precision• I1001HIGH-BARO—high precision• I1001PREMIUM-BARO—premium precision• I1001REFERENCE-BARO- reference precision

2. Options
The range of optional features includes:• Leak test – Automatically measures leak rates in thedesired units/minute or units/seconds• Switch test/analogue output – Accurate calibration ofpressure switches/integration into older ATE applications

PACE 1001 from the rear

5. Physical accessories
Please order the following as separate line items:
Part numberDescription

IO-ADAPT-G1/4

Adapter G1/8 male to G 1/4 female

IO-ADDAPT-G1/8

G1/8 male to G1/8 female

IO-ADAPT-1/8NPT

Adapter G1/8 male to 1/8 NPT female

IO-ADAPT-1/4NPT

Adapter G1/8 male to 1/4 NPT female

Please state area of use for instrument set up:

IO-ADAPT-7/16UNF

Adapter G1/8 male to 7/16 – 20 UNF female

EuropeIO-ADAPT-AN4

Adapter G 1/8 male to AN4 37 Deg maleNorth America

IO-ADAPT-AN6Adapter G 1/8 male to AN6 37 Deg male

IO-ADAPT-BARB

Adapter G 1/8 male to 1/4 hose

IO-ADAPTOR-KITContains one of each of the above adapters

IO-ADAPT-9/16AC

Adapter 9/16 18 UNF autoclave female to 1/8 NPTfemale

UK

IO-SNUBBER-1

Snubber reference port

Japan EU USA

IO-DIFF-KIT-LP

China

Differential connection kit low pressure: Helps
reduce the impact of thermal and/or pressurechanges in ambient conditions occurring during
the measurement cycle

IO-RMK-P100019” rack mount kit

Australia/New Zealand

IO-PAN-P1000

19” panel mount kit

4. PACE barometer unit selection

6. Supporting services

Please state unit selection.

Services ordering information:• Switch test/voltage free contacts – Accurate calibrationof pressure switches/automatically triggering ancillarydevices

3. PACE chassis – Area of use/mains leadJapan Asia
Rest of the world
Please state area of use for mains lead:

South Africa/Indiambar

psikPaPlease order the following as separate line items:

750-1150 mbar a

10.9 – 16.7 psi a

75-115 kPa aCalibrationExternal IDOS universal pressure modul

Part numberDescription

UKAS

PACE 1000 accredited pressure calibration

Related products

Druck manufactures a wide range of pressure transducers,transmitters, indicators, calibrators, controllers and air data
test systems. Our range of portable calibrators also covertemperature and electrical parameters.
Please refer to www.bakerhughesds.com/measurementsensing/pressure-measurement-and-calibration/test-andcalibration/pressure-controllers-pace for further information.

PACE family

PACE – pressure calibration and test solutions
PACE 1000 – Precision pressure indicator
PACE 1001 Barometer – Precision barometric indicator andrecorder
PACE 5000 – Single channel pressure controller chassis
PACE 6000 – Dual channel pressure controller chassis
CM0 – Standard precision high speed pressure controller
moduleCM1 – High precision high speed pressure controller module
CM2 – Premium precision high speed pressure controller
moduleCM3 – Reference precision high speed pressure controller
moduleCopyright 2023 Baker Hughes Company. All rights reserved.

920-623G

BHCS38696A(09/2023)

druck.com