MODEL 4610 ACCELEROMETERSPECIFICATIONSDIMENSIONSMEMS Accelerometer
DC Response, Ultra-Stable
Advanced Temp Compensation
Signal Conditioned Output
5,000g Over-Range Protection

The Model 4610 is an ultra-stable MEMS
accelerometer ideal for static and dynamic applications.
Theaccelerometer
offersintegraltemperaturecompensation with dynamic range from ±2 to ±200g. The
model 4610 incorporates a gas damped MEMS elementwith mechanical overload stops that provide shockprotection to 5,000g. The accelerometer has an operating
temperature range of -55°C to +125°C.

FEATURES


±2g to ±200g Dynamic Range
5000g Shock Protection-55°C to +125°C Operating Range
8 to 36Vdc Excitation Voltage
Gas Damping, DC Response
Integral Cable
Temperature Compensated

APPLICATIONSSENSOR SOLUTIONS /// Model 4610 Rev E

Low Frequency Monitoring
Transportation
Test & Instrumentation
Machine Control
Road Vehicle Testing

9/2015

Page 1

MODEL 4610 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)Residual Noise (µV RMS)
Spectral Noise (µg/Hz)

±2
1000
0-250
700
±0.5
<30.7
5000
600
38

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Output Resistance (Ω)
Full Scale Output Voltage (V)Insulation Resistance (MΩ)
Turn On Time (msec)
Ground Isolation±508 to 36<52.5<100±2>100<100Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Storage Temperature (°C)
Housing (Active Element & Electronics)
Humidity (Housing)±0.004±0.010-55 to 125-55 to 125
Hermetic Solder Seal
Epoxy Seal, IP65

TypicalTypicalPHYSICAL
Case Material
Cable
Weight (grams)
Mounting
Mounting TorqueAnodized Aluminum
5x #30 AWG Conductors PFA Insulated Leads, Braided Shield, TPE Jacket
6 (cable not included)
2x #4 or M3 Screws
6 lb-in (0.7 N-m)

Notes
±54000-700
800±0.5<30.75000
750
71±102000-1000
1000±0.5<30.75000
800
126±201000-1000
1500±0.5<30.75000
1200
379±50400-1000
4000±0.5<30.75000
800
632±100
200-1000
6000
±0.5
<30.7
5000
800
1265

±200
100-1000
8000±0.5<30.75000
800
2530

±5%1
<1 Typical
Passband
Passband@100Vdc

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit 1

Supplied accessories:

AC-A02285

2x #4-40 (7/16 length) Socket Head Cap Screw and Washer

Optional accessories:

AC-D02669
121

Triaxial Mounting Block
3-Channel Precision Low Noise DC Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 4610 Rev E9/2015

Page 2

MODEL 4610 ACCELEROMETER

2g Range
5g Range
10g Range
20g Range
50g Range
100g-500g Range

Typical Frequency Response
40353025Sensitivity Deviation (%)

20151050-5-10-15-20-25-30-35-40
101001000

10000

Frequency (Hz)2g Range
5g Range
10g Range
20g Range
50g Range
100g Range
200g Range

Typical Phase Response
4035302520Phase (degrees)

151050-5-10-15-20-25-30-35-40
10100100010000Frequency (Hz)SENSOR SOLUTIONS /// Model 4610 Rev E

9/2015

Page 3

MODEL 4610 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range+Cable Length4610-GGG-ZZZ-Cllll_____Cable (060 is 60 inches)
l_________Range (020 is 20g)
Example: 4610-020-060-C
Model 4610, 20g, 60” (5ft) CableNORTH AMERICAEUROPE

ASIA

Measurement Specialties, Inc.,a TE Connectivity Company
1000 Lucas Way
Hampton, VA 23666
Sales and Customer Service
Tel: +1-800-745-8008 or+1-757-766-1500
Fax: +1-757-766-4297
t&m@meas-spec.com

MEAS France SAS
a TE Connectivity Company
26 Rue des Dames
F78340 Les Clayes-sous-Bois
FranceSales and Customer Service
Tel: +33 (0) 1 79 33 00
Fax: +33(0)1 34 81 03 59
t&m@meas-spec.com

Measurement Specialties (China), Ltd.,a TE Connectivity Company
No. 26 Langshan Road
Shenzhen High-Tech Park (North)Nanshan District, Shenzhen 518057
China
Sales and Customer Service
Tel: +86 755 3330 5088
Fax: +86 755 3330 5099
t&m@meas-spec.com

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 4610 Rev E

9/2015

Page 4