COMMTEST220 AND COMMTEST240
Portable Data Collectors
Datasheet

Bently Nevada Machinery Condition Monitoring

109M0792 Rev. W

Description
COMMTEST220 and COMMTEST240 are portable data analyzer
solutions designed to monitor machinery in non-hazardous
environments. These small form-factor, rugged devices takevibration recordings and upload the collected data for detailed
analysis in System 1.
This is a two-part solution. The COMMTEST data acquisition
module uses Bluetooth to communicate with a wireless,industrial handheld device running the S1 Collector app.
Vibration recordings are viewed and stored on the handheld
device. The handheld runs the Android OS or Apple iOS, so you
can also use it like any smartphone or tablet to take pictures andvideo, send text messages, check email, make phone calls, track
your location, create SOS alerts, and run other enterprise apps.
The high performance data acquisition module provides long
battery life, yet is light & rugged. It hangs out-of-the-way on ashoulder strap, allowing you to focus on your job and movesafely between machines. The COMMTEST220 can sample two
sensor channels simultaneously, while the COMMTEST240
supports up to four channels. Both also support tachometer orKeyphasor triggering. These data acquisition modules provides
up to ten hours of continuous use, while supporting an Fmax up to
80 kHz and 12,800 lines of spectral resolution.
The System 1 software, paired with COMMTEST220 and
COMMTEST240, enables you to track all of your machinery using
a single enterprise monitoring system. Use System 1 to prepare
routes, analyze data, detect long-term trends, and set anddetect alarms. Then transfer the routes between System 1 andyour COMMTEST device using WiFi, file transfer, cellular network,
or a USB cable.

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

Hardware Features
l

l
l

l

l

l

l

l

l
l

l

l

Battery life of 10 hours allows the datacollector to run all day.
Five year warranty.
Rugged, water, and dust resistant (IP-65),
drop-tested.
Easy to carry: lightweight handheld unit,
shoulder strap for data collector, keepers
for sensors.
Safety breakaways on both shoulder
strap and sensor cables.
Upgradable industrial handheld
smartphone.
Handheld Push-to-Talk(PTT) button is
configured to start measurements(where available).
Up to four channel simultaneousrecordings (including on-route).
COMMTEST240 supports Triaxial sensors.
Up to 12,800 lines of spectral resolution,
80 kHz Fmax.
Supports accelerometer, velocity,
displacement, and voltage-output
sensors of AC and DC coupling types.

l

l
l
l

l

l

l
l

l

l
l

l

Modal Impact Testing (requires optional
instrumented hammer).
Set machine speed from spectral peak.

The COMMTEST portable connects via Bluetooth
to the handheld, which you can then sync with
System 1.

Software Featuresl

109M0792 Rev. W

Seamless on-and off-route datacollection.
Route overview & easy route navigation.
Fast sensor settling and data collection.
Patented 6Pack time-saving
measurement type.
Synchronous sampling, peak demod, andbump test.
Coast-down, run-up, and interval
capture support.
Custom note and numeric data entry.
Rotor Balancing, dual plane, reports
transferred to System 1.
Data quality warnings and easy-tounderstand alarm annunciation.
Supports operator driven reliability (ODR).
Cross Channel Phase measurement &reports.
Cross Channel FRF measurement &export UFF.

2/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

109M0792 Rev. W

Tachometer

Data Collector Specifications

Laser sensor powered byPortable Data Collector.

Sensors
Sensor input
Simultaneous
sampling

COMMTEST220: 2 channels.

Compatible
sensor types

Accelerometer, velocity,
displacement, and voltage.

AC coupledrange

16 V peak-peak.
Allows for ± 8 V sensor
output swing (± 80 g).

DC coupled
ranges0 to 20 V, -10 to 10 V, -20 to 0
V

Connectors

2 x LEMO connectors for CH1& CH2+
1 X LEMO connector for
Tach/Keyphasor
COMMTEST240 includes a CH
2.4 breakout cable.

Analog todigitalconversion
Sensorexcitation
current

Sensordetection

Optional
laser TachCOMMTEST240: 4 channels.

Triggers when beam onreflective tape range 10 cm to2 m.
Speed 6 to 250 krpm.

Other sensor
types
supported

Contact, TTL Pulse, Keyphasor

Power supply
to sensor

5 V, 40 mA

Input
100 KΩ
Impedance &± 28 V max
max values

24-bit ADC
0 mA or 2.2 mA(configurable), 24 V
maximum
2.2 mA powers IEPE/ICP-type
sensors
Measures sensor bias
voltage.
Warns if short circuit or notconnected.

3/14

Waveform
display± 20 V range

Threshold± 20 V range ± 0.4 V accuracy

Autothresholdmode

Signal ≥ 3 V pk-pk
Machine speed ≥ 60 rpm (1 Hz)
Settles within < 15 sHysteresis

0.2 V to 2.5 V user configurable

Speed range

Minimum: 3 rpm (0.05 Hz)
Maximum: 1,200,000 rpm (20kHz)

Signal pulse
width

10 µs minimum 2% duty cycle(for auto-threshold)Accuracy± (0.1 % + 1 cpm) for speeds up
to 5 kHz (300 kcpm).
± (0.3 %) for speeds above 5
kHz (300 kcpm).

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

Parameter Indication
Dynamic
signal range> 95 dB typical at 400 line
resolution

Harmonic
distortion

Less than -70 dB typical.
Other distortions and noise
are lower.

Unitsg or m/s2
in/s or mm/s
mil or mm or µm
peak, or peak-peak, or RMS

Magnitude &cursors

Overall value: RMS, derived pk,
derived pk-pk Dual cursors
Digital readouts on chart

Baseaccuracy

109M0792 Rev. W

Fmin possible
range

0 to Fmax.
Instrument zeroes all spectral
lines below Fmin.

Resolution(Spectrallines)

400; 800; 1,600; 3,200; 6,400;
12,800 lines
Channels

220
max lines

240
max lines

1

6,400

12,800

2

3,200

6,400

3 or 4N/A

3,200

Frequency
scale

Hz, CPM, Orders.
Linear scale with zooming.

For AC signal: ± 1% of readings
For DC level: ± 1% of full scale(1% is approximately 0.1 dB)

Amplitude
scale

Acceleration, velocity,
displacement.
Linear auto scaling.

Highfrequency
attenuation≤ 0.1 dB > 100 Hz to 10 kHz≤ 3 dB > 10 kHz to 40 kHz
Attenuation tolerances are in
addition to base accuracy.

Window
shapesHanning, Rectangular

Overlap

AC coupling
attenuation≤ 0.1 dB 10 Hz to < 100 Hz≤ 3 dB 1 Hz to < 10Hz

(0, 12.5, 25, 37.5, 50, 62.5, 75,87.5) % Dependent on Fmax
and number of lines
1, 2, 4, 8, 16, 32, 64, 128
Increases sampling time
proportionallyAttenuation
due toIntegration≤ 0.1 dB 10 Hz to < 100 Hz≤ 1.5 dB 1 Hz to < 10 Hz.
Values apply to single
integration (acceleration tovelocity). Double the values fordouble integration(acceleration todisplacement).

Number of
averages
Averagingtypes

Linear, exponential, peak hold

Demodulation
bandwidths

23 bandwidth options from125 Hz to 1,250 Hz and up to 16
kHz to 20 kHz.
Peak demod: 500Hz-40kHz, 140kHz, 2-40kHz, 5-40kHz

6Pack

Up to 40 kHz & 3,200 lines (1channel).
Up to 20 kHz & 1,600 lines (2-3channels).

Spectrum Display
Fmax ranges

25; 50; 100; 125; 150; 200; 300;
400; 500; 600; 800; 1,000;
1,200; 1,600; 2,000; 2,500;
3,000; 4,000; 5,000; 6,000;
8,000; 10,000; 15,000; 20,000;
30,000; 40,000; 60,000;
80,000 Hz or equivalen. CPM
values or orders-based from
1X to 999X.

Spectrum and waveform foreach of low-frequency, highfrequency, demodulation

4/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

Waveform Display

Storage

Number of 1,024; 2,048; 4,096; 8,192; 16,384;
samples
32,768Channels

Time
scale

220
max
samples

1

16,38432,7682

8,192

16,3843 or 4

N/A

8,192

Bluetooth

UFF Export

BT 4.0 Classic (s/n ≤ 101599)
BT 5.0 BLE (s/n ≥ 101600)

Battery and Charger
Battery type

Advanced Analysis

Modal
ImpactTesting

Balance jobs stored on
handheld and transferable
to System 1

Communication

240
max
samples

10 ms to 256 seconds or orders
based from 1 to 999 revs

Cross
channelspectrum

109M0792 Rev. W

OperatingTime

1 Reference and up to 3 othersensors

Charger typeCalculates Coherence andFrequency Response Function(FRF)

Charge rate

CH1 for hammer, up to 3
response channels, ≤ 10 kHz

Custom Lithium Ion pack, 7.4 V,
3920 mAh.
Not user accessible but
factory replaceable.
10 hours typical on-route datacollection
Internal charging, automatic
control.
External power pack 12 V DC, 3
A output.
2 A nominal 3 hours forcomplete charge.

MechanicalCalculates Coherence and
FRF — Accelerance, Mobility,
and ComplianceSize

5.1” W x 5.1” L x 1.4” H(130 mm W x 130 mm L x 35
mm H

Export of FRF data in Universal
File Format, for import into
Modal Analysis/ODS software.

WeightBalancing

1.3 lb ≈ 590g

Environmental Limits

Planes

2 planes, 2 sensors

Speed Range

3 to 12,000 RPM

Measurement
Type

Acceleration, Velocity,
Displacement

Weight modes

Angle 0° to 360°
Fixed Positions

Remove Trial
Weights

Yes/NoAutomatic recalculation

Operatingtemperature
Ambient
temperature
for charging
battery

5/14-4°F to 140°F (-20 to 60°C)
32°F to 77°F (0°C to 25°C)

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

Storagetemperature& humidity

Ruggedness
drop &
sealing

App Updates-4°F to 140°F (-20°C to 60°C),
95%relative humidity If
storage exceeds one month:
up to 95°F (35°C), 85% relative
humidity.

The Bently Nevada development teamperiodically updates the S1 Collector App withnew features and enhancements. Built into theapp is the new firmware to also automatically
update the SCOUT200 data collector. These
updates are free to all SCOUT200 owners who
have a current Maintenance and Support
agreement (M&S) for their System 1 software.

Drop spec MIL-STD-810F516.5-IV 26 drops at 4' (1.2 m)
onto concrete
IP65

The updates can be obtained from either ofthese locations:

S1 Collector App

1. Flexnet Licensing Portal > System 1 >System 1 Collector App.

Android 5.1 to 12
Operating
SystemLanguages

2. Google Playstore > Search "S1 Collector"

iOS 12 to 16(with s/n ≥ 101600 only,
requires BT 5 BLE).
Chinese, English, Finnish,
German, Japanese, Russian,
Spanish, Turkish

Cyber Security
The COMMTEST220 and COMMTEST240
solutions include an industrial handheld device
that runs the Android or Apple iOS operating
system and uses USB, cellular, Bluetooth and
Wi-Fi for communication. These solutions havefeatures that reduce but do not eliminate cybersecurity risks.
We strongly recommend implementing
additional cyber security protections such as:nnnn

109M0792 Rev. W

Controlling user access
Scanning for viruses
Applying timely updates
Using mobile device management
software
The recommended measures are atyour discretion and responsibility.
Some residual risk may remain.

For details, see the SCOUT200_Series Cyber
Security Hardening Guideline (document
131M1304) at BNTechSupport.com.

6/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

109M0792 Rev. W

Handheld Specifications Summary
A handheld device is not included in the COMMTEST kit. To order an Android device, add it as
an accessory item, or obtain an appropriately rated device directly from the supplier.
For self-supplied handheld devices, we cannot guarantee COMMTEST200 compatibility withevery make/model or version of Android. Contact BN TechSupport for a list of tested models,or to request a test.

Smart Ex 02 DZ2Tab ExD2 & DZ2

Display & Touch

5”, Gorilla glass, glove touch &wet mode8” 1920 x 1200, 480 nit, glove touch & wetmode

Wireless & GPS

4G/LTE dual SIM,
WiFi 2.4 & 5 GHz, BT 4.2, NFC, GPS

WiFi 2.4 & 5 GHz, BT 5.0,
NFC, GPS

Cameras12 MP rear, 5 MP front

13 MP rear, 5 MP front

OS

Android 11 or later

Android 11 or later

CPU

Qualcomm Snapdragon 450

Exynos 9810, 2.7 GHz,
Octa-core

Memory3 GB RAM, 32 GB ROM
up to 128 GB microSD

4 GB RAM, 48 GB internal,
up to 1 TB microSD

Battery

Replaceable Li-Ion 4400 mAh

Li-Ion 3.8V, 5050 mAh,
up to 11 hrs

Size & Weight

163 x 82 x 22 mm,
380 g

214 x 127 x
10 mm, 610 g

Ruggedness

IP68, MIL-STD-810G,-20 to 60°C

IP68, 1.2m drop,-20°C to +55°C

Warranty

3 years

3 years

Complete Specs,
Accessories

www.ecom-ex.com

www.ecom-ex.com

Device Image

7/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

Smart Ex 02 DZ2

109M0792 Rev. W

Tab ExD2 & DZ2
These handhelds are WIFI
models, so they do notsupport USB Tethering. This
prevents them from using USB Directcomms with
System 1.

SCOUT200
Compatibility

Available comms methods:
WiFi, File mode, Remotecomms.
To use USB Tethering, order
WWAN version direct from
Pepperl & Fuchs.
Numerous accessories are available for these handhelds, direct from each handheld
manufacturer (e.g. charging stations, belt-clips for the smartphones, and cases for thetablets). You can find the accessories pages on the handheld manufacturers websites.
If Apple iOS handhelds are preferred, these are not available through Bently Nevada andshould be purchased by the end user. If comms via USB tethering are used, the handheld
must have an active SIM card installed (i.e. a WiFi-only iOS tablet cannot communicate via
USB cable).

8/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

109M0792 Rev. W

RED/EMC

Compliance and Certifications
This section applies only to the Data
Collector, not the handheld device.

EN 61000-6-2
EN 61000-6-4
EN 300 328
EN 301 489-1
EN 301 489-17
EN 61326-1
2014/53/EU
2014/30/EU

This device complies with Industry Canada
license-exempt RSS standard(s) and complies
with part 15 of the FCC Rules, Class A device.
Operation is subject to the following two
conditions:ROHS

2011/65/EU
The overall hazardous system rating
is defined by the minimum rating of
the data collector, industrial
handheld, and accessories.

1. This device may not cause harmfulinterference.
2. This device must accept any interference
received, including interference that may
cause undesired operation.
Changes or modifications not
expressly approved by Bently Nevada
could void the user's authority tooperate the equipment.
This equipment generates, uses, andcan radiate radio frequency energy
and, if not installed and used inaccordance with the instructions,
may cause harmful interference toradio communications. However,there is no guarantee thatinterference will not occur in aparticular installation.
For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).
For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

9/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

Ordering Information
For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).
For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

COMMTEST220-AA
A: Data Collector Approval Options
00
Part Number

None

109M0792 Rev. W

Part Number

Description

Qty

AS3100S2Z0 or
200350*Accelerometer, side exit

1

151M8244-01 Accelerometer cable,
coiled, safety breakaway

2

113M5585

Accelerometer magneticbase

2

109M2385012

BNC to silver-LEMO cable.
Connects rack buffered
outputs to CH1/CH2 inputs.

2

110M8172012

BNC to black-LEMO cable.
Connects rack buffered
Keyphasor to Tach input.

1

117M1446

Belt Clip1

109M238401

Neck strap with Sensor
Keeper1

112M4551

Hard carrying case1

109M1269

SCOUT200 Series User
Guide

1
1

Description

Qty

S1 Collector app, digitaldelivery (see App Updates
on page 6)

1

COMMTEST220 Data
Collector

1

109M1270-01 SCOUT200 Series Quick
Start Guide

108M4044

AC power adapter

1

PLUS0230

Category A power plug
(USA / Canada)

1* Kit items with multiple part numbers listed have
limited regional availability due to certification
requirements.

PLSA0241

Category D power plug
(South Africa / India)

1

PLAU0228

Category M power plug
(Australia / New Zealand /China)

1

PLHK0245

Category G power plug
(Hong Kong / UK)1

PLEU0229

Category C Power plug
(Europe)1AM3100T2Z0 or
200350*Accelerometer, top exit

1

The COMMTEST220 kit does
NOT include a handheld device, nor a
System 1 software license. They areordered separately under Optional
Accessories below.
Metrology certification can be requested whenan order is placed, however this service ischarged. Standard test data can be requested
for free, but is not evidence of calibration.
To request a periodic metrology calibration,
contact Bently Nevada Tech Suppport.

10/14

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

COMMTEST240-AA

Part Number

A: Data Collector Approval Options
00
Part Number

None
Description

Qty

S1 Collector app, digitaldelivery (see App Updates
on page 6)

1

COMMTEST240
Data Collector

1

108M4044

AC power adapter

1

PLUS0230

Category A power plug
(USA / Canada)1

PLSA0241

Category D power plug
(South Africa / India)

1

PLAU0228

Category M power plug
(Australia / New Zealand /China)

1

PLHK0245

Category G power plug
(Hong Kong / UK)1

PLEU0229

Category C Power plug
(Europe)1AM3100T2Z0 or
200350*Accelerometer, top exit

3

AS3100S2Z0 or
200350*Accelerometer, side exit

1

Description

4

Accelerometer magneticbase

4

109M2385012

BNC to silver-LEMO cable.
Connects rack buffered
outputs to CH1/CH2 inputs.

2

110M8172012

BNC to black-LEMO cable.
Connects rack buffered
Keyphasor to Tach input.

1

Qty

113M9350

Channel 2-4 breakout
cable

1

117M1446

Belt Clip1

109M238401

Neck strap with Sensor
Keeper1

112M4551

Hard carrying case1

109M1269

SCOUT200 Series User
Guide

1

109M1270-01 SCOUT200 Series Quick
Start Guide

1* Kit items with multiple part numbers listed
have limited regional availability due tocertification requirements.
The COMMTEST240 kit does
NOT include a handheld device, nor a
System 1 software license. They areordered separately under Optional
Accessories below.
Metrology certification can be requested whenan order is placed, however this service ischarged. Standard test data can be requested
for free, but is not evidence of calibration.
To request a periodic metrology calibration,
contact Bently Nevada Tech Suppport.

Optional Accessories
Hand Held Devices

151M8244-01 Accelerometer cable,
coiled, safety breakaway
113M5585

109M0792 Rev. W

11/14

146M8212-01

Smart-Ex 02 DZ2(North America &Europe)175M2586-01

Tab-Ex D2 North
America

145M3278-01

Tab-Ex DZ2 North
America

146M8220-01

Tab-Ex DZ2 Europe

COMMTEST220 AND COMMTEST240 Portable Data Collectors
DatasheetPart Number

Hand Held Devices
To obtain a handheld device
suitable for a location not listed
above, contact a handheld device
distributor within the destination
country.

Other Accessories
3071/01

System 1 software and license

108M4069

Laser Tach powered by
COMMTEST220 and
COMMTEST240.

5-meter cable for laser Tach
108M4069.

122M9408

Monarch Compact Smart Laser
Sensor (CSLS). Higher powered(3 mW) but larger size (2.6" x2.4" x 2"). Order cableseparately.

125M209901

Cable for CSLS laser, 1.5 m

113M5529-01

Reflective tape
One roll, 60 cm (23.7 in)

VBMR0222

Pair of safety rings for attaching
strap to data collector

138M7749

Accelerometer cable, 4 ft,
straight, safety breakaway

TP500

Accelerometer, top exit, 500
mV/g, 10 g pk, 0.2 Hz - 14 kHz

Accessory Kits
Balancing Kit - 161M0958

113M5529-01 Reflective tape One roll,
60 cm

Qty
1

12/14

Qty

LASA0315

Laser Tach Kit Zone 2
rated

1

CBL50216

Laser cable Five meters

1

MAGA0063

Laser magnetic stand1

159M9583*

Sensor extension cable,
5m

2

CBBL0026

Carrying case for the kit

1substituted with the CB5G0024 and CB5R0025
cables and two 161M2913 adapters.

CBL50216

Description

Description* Depending on availability, this cable may be

Order cable separately.

Part Number

109M0792 Rev. W

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

109M0792 Rev. W

157M0806-01

Zone 2 Laser Tach Kit - LASA0315
Part
NumberDescription

108M4064

Laser Tacho Holder

1

108M4066

Circlips - 20Mm Stainless1

108M4067

Arp115 Oring

2

108M4069

Laser Tach Zone 2 rated

1

Part Number

Qty

Impact_Hammer_Kit - AA-BB-CC-DDEE-FF

156M6189

Triaxial sensor,
100 mV/g ± 20%,
0.5 Hz .. 5 kHz,
8 µg/√Hz noise @ 10Hz

157M0852-01

Cable for SCOUT200 to triax
156M6189,
1.5 m, straight,
safety breakaway

156M6204

Magnet, 2 pole, 38 mm
diameter,
32 mm high, 25 kg pull,¼-28 female thread

A: 500lbf pk, 10mV/lbf, 0.3lbm (285570-01)
00

None

01

Hammer Included

Description

157M0809-01

B: 1000lbf pk, 5mV/lbf, 0.3lbm (285570-02)

Part Number

00

None

155M3055

01

Hammer Included

Triaxial sensor,
100 mV/g ± 10%,
0.6 Hz .. 10 kHz,27 µg/√Hz noise @ 10 Hz

156M6180

Cable for SCOUT200 to triax
155M3055,
1.5 m, straight,
safety breakaway

154M1355

Magnet, 2 pole, 35 mm
diameter, 22 mm high,
23 kg pull, ¼-28 female thread,
triax alignment notch

C: 5000lbf pk, 1mV/lbf, 2.4lbm (285570-03)
00

None

01

Hammer Included

D: N/A
E: N/A
F: N/A

Triax Sensor Kits

Description

Language Specific Warning Labels

Triaxial sensor kits are certified forhazardous area use in limited regions(NAM, EU, etc). Other regions mayneed to source locally. Contact
TechSupport for suggested
manufacturer part numbers.

13/14

110M9913

Magnet base label

110M9914

Sensor keeper side label

110M9915

Sensor keeper front label

111M7144

Hard carrying case label

COMMTEST220 AND COMMTEST240 Portable Data Collectors
Datasheet

109M0792 Rev. W

Copyright 2024 Baker Hughes Company. All rights reserved.

Bently Nevada, COMMTEST, Keyphasor, M2, SCOUT, System 1 and Orbit Logo are registered trademarks of Bently
Nevada, a Baker Hughes business, in the United States and other countries. The Baker Hughes logo is a trademark of
Baker Hughes Company. All other product and company names are trademarks of their respective holders. Use of
the trademarks does not imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

14/14