DPS5000 SDI-12

Digital pressure sensing platform
The DPS5000 SDI-12 from Druck, part of the UNIK5000
family, offers integrated digital electronics to enhance
the performance level of the UNIK 5000 Pressure Sensing
Platform to levels unmatched by traditional analogue
sensors. It uses SDI-12 protocol, over which fully compensatedreadings of pressure and temperature are sent, as well as
control of many functions of the device.

Features

High quality• Customer set filter settingWith 50 years of pressure measurement experience, ourfield-proven Druck technology is at the heart of the newplatform, resulting in a range of high quality, high stability
pressure sensors.

Expertise
We have the people and the knowledge to support yourneeds for accurate and reliable product performance; ourteam of experts can help you make the right sensor selection,
guiding you and providing the help and tools you need. It
is important that you ensure that the sensor materials andperformance selected are suitable for your application.

• Ranges from 350 mbar to 70 bar• Total accuracy to ±0.1 % FS
• Stainless steel or titanium construction• Wide temperature range -40 °C to 80 °C• Low power• Excellent long-term stability• Customizable and reconfigurable output formats

DPS5000 SDI-12 specifications
Operating pressure ranges

Physical specifications

Gauge ranges

Environmental protection

Any zero based range from 350 mbar to 70 bar (6 to 1000 psi)

Up to IP68, see electrical connector section

Absolute ranges

Operating temperature range

Any zero based range from 350 mbar to 70 bar (6 to 1000 psi)

-40 °C to 80 °C (-40 °F to 176 °F)

Barometric ranges

Pressure media

Any upper range limit (URL) with a minimum span of 350
mbar (5 psi) between 700 mbar to 1300 mbar (10 to 18 psi)

Fluids compatible with stainless steel 316L and Hastelloy C276
or titanium

Over pressureStatement in accordance with the European pressure
equipment directive

2 × pressure range for barometric range
4 × pressure range for all other ranges
Containment pressure
6 × pressure range (200 bar (3000 psi) maximum)
SupplyVoltage: 6 Vdc to 30 Vdc
Current: <10 mA measure at 25 °C
15 mA peak
<50 μA standby
Output/communications
SDI-12 protocol V1.4
Power on time1000 ms to acquisition from power on
Insulation resistance

Enclosure materials
Stainless steel 316L or titanium (body), PVDF (nose cone), andcable sheath as specified
Pressure connectors
See the ordering information
Electrical connector
Option code

Description

IP rating

2Raychem cable

IP65

3

Polyurethane depth
cable

IP68

4

Hytrel depth cable

IP68

Note: Maximum cable length is 457 m (1500 ft).

Wiring details
Core color

Function

BlackCasePerformance

Red+ VE Supply

Pressure performanceBlue- VE Supply

Accuracy over the calibrated temperature range including
zero and span setting and the effects of non-linearity,hysteresis and repeatability

WhiteSDI-12 Data

ScreenCase>100 MΩ at 500 Vdc

A3 Premium: ±0.1 % FS over -10 to 50 °C (14 to 122 °F)
A2 Improved: ±0.2 % FS over -40 to 80 °C (-40 to 176 °F)
For barometric ranges FS is the URL
Accuracy effects increase pro-rata for pressure ranges
below 700 mbar
Temperature performance

General certifications
RoHS 2011/65/EU
CRN Certified 0F18611.513467890YTN for pressure ranges up toand including 100 bar
CE conformity
Pressure equipment directive 2014/68/EU: soundengineering practice

Accuracy over the calibrated temperature range ±3 °C (6 °F)
with 0.1 °C (0.2 °F) resolution

EMC directive 2014/30/EU, AUS/NZ EMC compliant

Long term stability

BS EN 61000-6-1: 2007

±0.05 % FS/year typical±0.1 % FS/year maximum at reference conditions±0.2 % maximum over five years at reference conditions isavailable by arrangement with the factory
Stability effects increase pro-rata for pressure ranges
below 700 mbar

BS EN 61326-1:2013
BS EN 61000-6-3: 2007 + A1:2011

Ordering information1) Select part number

Main product FamilyDPSDigital output transducer
55000 series		Diameter and material		
025mm stainless steel		T25mm titanium			Electrical connection			
2Raychem cable (See Note 1)			3
Polyurethane depth cable			4Hytrel depth cable				Electronics option				ESDI-12 protocol					Compensated temperature range					
TA-10 °C to +50 °C (See note 2)					
TC-40 °C to +80 °C (See note 3)						Accuracy						A2Improved						A3Premium							Calibration							CCFull thermal								Hazardous area certification								H0None									Pressure connector									- Stainless steel									PAG1/4 female									PBG1/4 male Flat									PCG1/4 male 60o Int cone									PDG1/8 male 60o Int cone									PE1/4 NPT female									PF1/4 NPT male									PG1/8 NPT male									PHM20 x 1.5 male (3mm bore)									PJM14 x 1.5 60o Int cone									PL7/16-20 UNJF male 74o Ext									PNG1/2 male									PQG1/2 quick connect									PR1/2 NPT male									PV7/16-20 UNF female									PWDepth cone (G1/4 female)									RVR1/4 male									- Titanium									PAG1/4 female									PBG1/4 male flat									PF1/4 NPT male									PHM20 x 1.5 male (3mm bore)									PNG1/2 male									PR1/2 NPT male									PWDepth cone (G1/4 female)									RJM20 x 1.5 male (8mm bore)									RKM20 x 1.5 male (14mm bore)									    Sensor output: can be selected in any order (see note 4)									    - Level									    L1   Meters									    L2   Centimeters									    L3   Feet									    - Temperature									    T1   Kelvino									    T2   Celsiuso									    T3   Fahrenheit									    - Pressure									    
P   Pressure (as specified)									    - Voltage (Power Supply)									    
V   Volts (as specified)									    - No Output									    
N   No output (see note 5)
DPS5  

* = Sensor output

0  

2  

E-TA-

A3-

CC-

H0-PA- *-*-*-*2) Specify maximum working pressure unit and reference. Available units are:Symbolbar

Description
barpsipounds/square inch

mmH2O
mH2O

hecto Pascal

ft H2Ombar
PahPa

millibar
PascalSymbolkPain H2O

Description
Kilo Pascal

SymbolmmHg

Description
millimeters mercury

meters water

kgf/cm2

Kilograms force/centimeters square

millimeters water
inches waterfeet water

inHg
atmMPainches mercury
atmosphere

Mega Pascal3) Specify cable length and units:

Integer values in m or ft (maximum length 457 m (1500 ft)).
Notes:
1. Not available with a titanium body.			
2. Choose A3-Premium accuracy with this option.		
3. Choose A2-Improved accuracy with this option.

4. Maximum of 4 outputs. Each output option can only be selected once.
5. Once No output (N) is selected, subsequent options default to N.

Typical order examples
DPS503E-TA-A3-CC-H0-PW-T2-L1-P-V 700 mbar gauge 10m cable
DPS5T4E-TC-A2-CC-H0-PA-L3-T3-N-N 10 bar absolute 10m cableMechanical drawings

Note: All dimensions in millimeters.

Copyright 2022 Baker Hughes Company. All rights reserved.

920-690B

BHCS38665 (03/2022)

druck.com