ADROIT6200

Pressure sensing platform
The ADROIT6200 is a high performance, robustpressure measurement device. It combines thebest mechanical properties of micromachined
silicon in a fully welded stainless steel 316L bodywith the latest digital processing capability tooffer levels of accuracy, previously unavailable
in a device of this type. It offers a full suiteof pressure measurement including gauge,absolute and differential references andpressure ranges from 68 mbar to 350 bar.

Digital and Analogue
By using digital signal processing, the highest
levels of performance over temperature areachieved. However, the final stage of processing
converts the signal back into a conventional
analogue output for easy interfacing to existing
infrastructure. Fast ASIC technology ensures
response times around 1 ms and fast switch
on for pulse power operation. The digitalprocessing also enables the replacement ofmechanical adjustment to potentiometers withautomatic calibration adjustment using an App.

Small and Robust
The highest-grade components are designed
to withstand high levels of shock, vibrationand temperature extremes. Components
are welded into a small, convenient, 19 mmdiameter package to offer best in classperformance in the harshest environments.

Druck.com

Druck Expertise
Druck have used 50 years of experience to design
the ADROIT6200. From the silicon processing,through the mechanical construction of thepressure sensing module and electronics design,
to the selection of electrical connectors, eachcomponent’s performance has been optimized
to meet your requirements. Our team of experts
can help you make the optimum productselection for your application.

Features•Performance to 0.02% span BSL•

Pressure ranges from 68 mbar to 350 bar(1 to 5000 psi)

•Gauge, Sealed Gauge, Absolute,
Barometric and Differential reference•

4-20 mA , 20-4 mA and configurable voltage output•

Total accuracy up to 0.1% span•Frequency response to 1 kHz•

Stainless steel 316L construction•

Operating temperature range up to -40 °C to +125 °Cand survival from -55 °C to +150 °C

Measurement

Operating Pressure Ranges
Gauge ranges (g)Zero and non-zero based ranges
Upper range Limit (URL): 68 mbar to 70 bar
Lower range limit (LRL): -1 to 35 bar
Span limit: for gauge depression ranges, Min URL is 0,
span needs to be greater than 68 mbar.

Sealed gauge ranges (sg)Zero and non-zero based ranges
Upper range Limit (URL): 10 bar to 350 bar
Lower range limit (LRL): -1 to 200 bar
Span limit: Span must be greater than 50 % of the URL

Absolute ranges (a)Zero and non-zero based ranges
Upper range Limit (URL): 344 mbar to 350 bar
Lower range limit (LRL): 0 to 175 bar
Span limit: Span must be greater than 50 % of the URL.
See Note 8 in ordering information.

Containment Pressure
Form

Pressure Range

ContainmentPressureGauge
Absolute
Barometric
Sealed Gauge
Differential (+ve port)≤70 bar

6 x FS>70 bar, ≤350 bar

6 x FS (700 barmaximum)Must not exceed
positive port bymore than 6 x FS
(15 bar maximum).

Differential (-ve port)

Electrical Parameters
Outputs•

4-20 mA•

20-4 mA•

0-5 Volts 3-wire non true-zero*•Configurable: 3-wire voltage output versions within
the range of 0 to 10 V with the following limitations:- Minimum span of 4 V- Maximum lower limit elevation equal to the span

Barometric (b)Examples:Upper range Limit (URL): 1.3 bar
Lower range limit (LRL): 350 mbar
Span limit: Span must be greater than 350 mbar

ValidInvalid

1 to 6 V
0.5 to 4.5 V

1 to 3 V (span too small)
6 to 10 V (offset too large)

Wet-dry differential (WD)
Zero and non-zero based ranges
Upper range Limit (URL): 68 mbar to 35 bar
Lower range limit (LRL): -1 to 17.5 bar
Span limit: Span must be greater than 50 % of the URL.
See Note 8 in ordering information.

Wet-wet differential (WW)
Zero and non-zero based ranges
Upper range Limit (URL): 344 mbar to 35 bar
Lower range limit (LRL): -1 bar to 17.5 bar
Span limit: Span must be greater than 50% of the URL

Over-pressure

The following pressure can be applied without causinga shift in calibrated accuracy:•

6 x FS for ranges up to 700 mbar•

2 x FS for barometric ranges•

4 x FS for ranges up to 70 bar•

4 x FS for ranges up to 350 bar (700 bar max)

For differential versions the negative side must notexceed the positive side by more than:•

6 x FS for ranges up to 150 mbar

•4 x FS for ranges up to 700 mbar

•2 x FS for all other ranges up to a maximum of 15 bar

Differential line pressure maximum 70 bar

Output voltage range can be specified to a resolutionof 0.1 V
The output will respond to at least 110 % of the appliedpressure* Non true-zero, the output will saturate at <50 mV.

Supply Requirements
4-20 mA output
12-28 Vdc

20-4 mA output
12-28 Vdc

Voltage output
7-28 Vdc <3 mA
Supply voltage must be 2 V greater than the outputvoltage selected.

Insulation

Greater than 100 MΩ at 500 Vdc.

Power On Time

From power on to a stable reading within specification
Less than 30 ms.

Performance Specification

Physical Specifications

Accuracy

Environmental protection

Including zero and span setting accuracy, NLH&R andthermal error:
Total Accuracy% of Span
Premium

0.10

Improved

0.20

NLH&R (Non-linearity
Hysteresis andRepeatability) at 23 °C

0.02 % Span BSL (BestStraight Line)
0.04 % Span BSL

Values increase pro-rata for pressure spans less than
1000 mbar and double for barometric ranges.

Response timeLess than 1 ms

Resolution

0.01 % FS span

Stability

Long-term stability 0.05 % Span/year typical(0.1 % Span/year max), increasing pro-rata
for pressure ranges below 1 bar.

Line pressure effects (differential versions only)•

Zero Shift <±0.03 % Span /bar of line pressure•

Span shift <±0.03 % Span /bar of line pressure•

Effects increase pro-rata for differential ranges
below 700 mbar

Orientation sensitivity
Gauge, absolute, and wet-dry differentials
Units are calibrated mounted pressure connection
down. Output will change by less than 1 mbar /g which
can be zeroed out during calibration
Wet-wet differentials
Units are calibrated with the positive port down. Output
will change by less than 5 mbar/g which can be zeroedout during calibration.

Vibration and shock
Sinusoidal Vibration to DO-160G Curve W. 5 to 2000 Hz,
30 g peak.
Random vibration to DO-160G Cat. R (robust) Curves
D1+E1. 10 to 2000 Hz, peak ASD 0.16 g2/Hz.
Random vibration to BS EN 61373:2010. 5 to 250 Hz, peakASD 6.12 g2/Hz.
Shock, 1000 g half-sine for 1 ms.

See Electrical connectors section.

Mass

60 g approx.

Operating temperature range
See Electrical connectors section.

Pressure media compatibility

Fluids compatible with stainless steel 316L
and Hastelloy C276.
Not suitable for media that has an oxygen concentration >21% orother strong oxidizing agents. This product contains materials or
fluids that may degrade or combust in the presence of strongoxidizing agents.

Enclosure materials

Stainless steel 316L body.
Connector typePolyurethane cable(metal crimp)
Raychem cable(metal crimp)
MIL-C-26482
M12x1 Male 4-pin
Micro-DIN (9.4 mmpitch)

Material for TB
temp range

Material for TD
temp range

Raychem

Raychem

PBT, Brass H62

Glass, Gold plated
nickelGlass, Alloy52
Not available

Polyurethane

Nylon 6, Brass H62
Nylon 66, Tin plated
Brass, Copper AlloyNot available

Pressure connectors•

PA: G1/4 Female•

PB: G1/4 Male Flat•

PC: G1/4 Male 60° Int Cone•

PE: 1/4 NPT Female•

PF: 1/4 NPT Male•PG: 1/8 NPT Male•

PJ: M14 x 1.5 60° Int Cone•

PK: M12 x 1 Male 60° Int Cone•

PS: 1/4 Swagelok Bulkhead•

PT: G1/4 Male Flat Long•

P14: M8 x 1 Male•P18: M10 x 1 Male•

P22: 7/16-20 UNF male 74° Ext Cone•

P33: 7/16 UNF Female•

P60: M12 x 1.5 Int. Cone•P61: G1/4 Female with Wire Lock•

RA: 1/4 VCR Female•

RC: G1/4 Male Flat Cross Bore•

RF: 1/4 VCR Male•

RQ: NW16 Flange•

RZ: 1/4-28 UNF LH THD 58° Cone

Pressure connection dimensions are detailed in
document “ADROIT6200 Pressure Connectors”
available on Druck.com.

CE Conformity

Electrical connectors
DescriptionPolyurethane cable(metal crimp)
Raychem cable(metal crimp)
MIL-C-26482(6 pin Shell size 10)
M12x1 Male
(4 pin Type A coded)
Micro-DIN(9.4 mm pitch)

IP RatingLocation

IP65

Operating Temperature
MinMaxIndoor-40 °C+80 °C

IP65

Indoor-55 °C+125 °C

IP65/67**

Outdoor-55 °C

IP65/67**

Outdoor-55 °C

IP65

Indoor-40 °CAs compensatedtemperature*
As compensatedtemperature*

MIL-C-26482
Alt WiringM12x1 Male 4-pin

M12x1 Male 4-pin
Alt Wiring•EMC Directive 2014/30/EU•

BS EN 61326-1: 2013:Electrical Equipment for
Measurement, Control andLaboratory Use•

BS EN 61326-2-3: 2013:
Particular Requirements for
Pressure Transducers•BS EN 50121-3-2: 2016+A1:
2019: Railway applications

Option CodePolyurethane Cable 1

Micro-DIN(9.4 mm pitch)

Pressure Equipment+80 °C

Connections

MIL-C-26482•

Engineering Practice**Note: IP65 for temperature range -20 °C to +80 °C and IP67 for temperature range -55 °C to+125 °C.

Raychem Cable

RoHS Directive 2011/65/EU
Directive 2014/68/EU Sound*Note: Units with a compensated temperature range up to 125 °C will withstand short periods
operating at temperatures up to 150 °C. Exposure to temperatures above 125 °C will shorten
operating life.

ElectricalConnection•26DEGW

Connections

4-20 mA

VoltageOutput

Red+ve Supply+ve Supply

Yellow-+ve Output

Blue--White-ve Supply

0V Common

Orange--Black--

Cable Screen

Case

Case

Red+ve Supply+ve Supply

White-+ve Output

Green--

Blue-ve Supply

0V Common

Black--

Cable Screen

Case

Case

A+ve Supply+ve Supply

B-ve Supply+ve Output

C--

D-ve Supply

0V Common

E--F--

1+ve Supply+ve Supply

2-ve Supply

0V Common

3-+ve Output

ECaseCaseA-+ve Supply

B-0V Common

C-+ve Output

D-

0V Common

E--F-

-

1+ve Supply+ve Supply

2-+ve Output

3-ve Supply

0V Common

4--

1+ve Supply+ve Supply

2-ve Supply

0V Common

3--4-+ve Output– Electromagnetic
compatibility – Rollingstock – Apparatus. 4-20 mA
version only.

Ordering Information

1. Select model number

Product Series
ADROIT6
ADROIT6200		
Diameter and Material		219 mm stainless steel			Electrical Connector			1
Polyurethane Cable (metal crimp)			2
Raychem Cable (metal crimp)			6MIL-C-26482 Bayonet 			Note 1			D
Micro DIN (9.4 mm Pitch) 		Note 2			EMIL-C-26482 Bayonet alternative wiring Note 1			G
M12 x 1 4-Pin			Note 1			W
M12x1 Male 4-Pin alternative wiring		
Note 1				Electronics Option				24 to 20 mA				40 to 5 Volts 3-wire				5Configurable Voltage 3-wire				R20 to 4 mA					Compensated Temperature Range					TB-20 to +80 °C					TD-40 to +125 °C		Note 3								Accuracy						A2Improved						A3Premium		 Note 4							Calibration							CD Total Precision and Zero and Span Data								Hazardous Area Approval								H0None									Pressure Connector									PA G1/4 Female														PB G1/4 Male Flat									PC G1/4 Male 60° Int Cone									PE 1/4 NPT Female									PF 1/4 NPT Male									PG 1/8 NPT Male									PJ M14 x 1.5 60° Int Cone									PK M12 x 1 Int Cone									PS 1/4 Swagelok Bulkhead Note 7									PT G1/4 Male Flat Long									P14 M8 X 1 Male
Note 7									P18 M10 X 1 Male									P22 7/16-20 UNF Male 74° Ext Cone									P33 7/16-20 UNF Female									P60 M12 x 1.5 Int. Cone
Note 7									P61 G1/4 Female with Wire Lock Note 5 Note 7									RA 1/4 VCR Female
Note 7									RC G1/4 Male Flat Cross Bore Note 5									RF 1/4 VCR Male
Note 7Note 6 Note 7									RQ NW16 Flange									RZ1/4-28 UNF LH THD 58° Cone Note 5 Note 7					ADROIT62

62-TB-A2-

CD

Note 1: Mating connector not supplied. (see Accessories, section 3).
Note 2: Mating connector supplied. (see Accessories, section 3).
Note 3: TD compensated range is only available for connector
options 2, 6, G or W.
Note 4: Premium Accuracy is not available if TD compensatedtemperature range is selected.

-H0-PA(Example Configuration)

Note 5: This connection is only available on pressure ranges
between ≥10 bar and ≤350 bar.
Note 6: This connection is only available on pressure ranges up to<10 bar.
Note 7: This connection is not available with differential
pressure ranges.

2. State pressure range and units: e.g. 0 to 10 bar, -5 to +5 psi Note 8
 nit options are:U
Symbol		Description
bar		barmbar		
millibarpsi		pounds/sq. inch
Pa		 Pascal
hPa		
hectoPascalkPa		
kiloPascal
MPa		
MegaPascal
mm watermmH2O		cm water
cmH2O		

Symbol		mH2O		
inH2O		ftH2O		mmHginHg		
kgf/cm2		
atm		
Torr		

Description
metres water
inches waterfeet water
mm mercury
inches mercury
kg force/sq. cm
atmosphere
torr

Note 8: For ranges greater than 10 bar the zero offset needs to be less than 50% of the span.

3. State pressure reference: e.g. gauge
Reference options are:•gauge•absolute•

barometric•

sealed gauge•

wet-dry differential•

wet-wet differential

4. Electrical Connector options 1 and 2: State cable length and units: Integer values only in ft or m.
•

Minimum cable length: 1 m (3 ft)•Maximum cable length: 3 m (10 ft)

5. Electronics option 5: State output at minimum and maximum pressure: e.g. output 0.5 to 4.5 V

Request a quote here: http://bit.ly/Adroit6000contactus

Accessories (to be ordered as separate line items)
1. ADROIT6200 Interface box part number: ADROIT-Interface

The interface is used with a Windows-based PC or an Android device (laptop, tablet or phone). It enables
the user to make small adjustment to the zero and span settings of the sensor for calibration purposes. It is
supplied with a USB lead to USB-C (Android device) or USB-A (Laptop).

2. Test leads

Set of 2 off 4 mm leads and crocodile clips for connection to the ADROIT6200 sensor are available.
Part Number: 209-359. 2 sets are required for calibration.

3. Mating connectors•

For MIL-C-26482 Bayonet 		Part Number 163-009•

For M12 x 1 4-Pin 			Part Number 149M7393-1•

For Micro DIN (9.4 mm Pitch)		

Part Number 192-257-01 (one supplied with each sensor)

4. Cable Assemblies
A made up electrical connector with a length of cable terminated in solder tinned wires.
(1) Select Part Number

Main Product
UNIKCABLE
Cable Assembly		Electrical Connector		6MIL-C-26482 Bayonet		D
Micro DIN (9.4 mm Pitch)		G
M12 x 1 4 Pin					Cable			1
Polyurethane Cable			2Raychem Cable			
UNIKCABLE- 6 -2 (Example Part Number)(2) State Cable length and units (Integer value only)

Minimum length 1 m (3 ft)
Maximum length 200 m (600 ft)
Example: UNIKCABLE-6-2 5 m

Mechanical Drawings
55

59
19 A/F

1617ADROIPRESSU25

ADROIT62##-T*********** AC
S/N *********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. LE

ADROIPRESSU

ADROIT62##-T*********** AC
S/N *********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. LE

LOW PRESSURE CONSTRUCTION(PRESSURE RANGES: <10 bar)

MEDIUM PRESSURE CONSTRUCTION(PRESSURE RANGES: ≥10 bar to ≤350 bar)

15.5 BOTH ENDS

5.5

ADROI
PRESSU72

ADROIT62##-T*********** AC
S/N *********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. LE

19MALE PRESSURE CONNECTION [2]1619191266

WET/WET & WET/DRY
DIFFERENTIALCONSTRUCTION

HIGH TEMP (TD) M12x1 4-PIN

20197

STANDARD TEMP (TB) M12x1 4-PIN

POLYURETHANE/RAYCHEM CABLE

3213191915.51933.517

MICRO DIN (9.4 mm PITCH)

STANDARD TEMP (TB)
MIL-C-26482 BAYONETHIGH TEMP (TD)MIL-C-26482 BAYONET

Copyright 2023 Baker Hughes Company. All rights reserved. This material contains
one or more registered trademark of Baker Hughes Company and its subsidiaries in
one or more countries. All third-party product and company names are trademarksof their respective holders.
BHCS39260		
920-699D(11/2023)

druck.com