UNIK 5000

Pressure sensing platform
The new UNIK 5000 is a high performance configurable
solution to pressure measurement. The use of
micromachined silicon technology and analogue circuitry
enables best in class performance for stability, low power
and frequency response. The new platform enables you toeasily build up your own sensor to match your own precise
needs. This high performance, configurable solution topressure measurement employs modular design and leanmanufacturing techniques.

High quality

selection, guiding you and providing the help and tools you
need. It is important to ensure that the sensor material andperformance selected are suitable for your application.

Features• Ranges from 70 mbar (1 psi) to 700 bar (10000 psi)
• Accuracy to ±0.04% Full Scale (FS) Best Straight Line (BSL)• Stainless steel construction• Frequency response to 3.5 kHz

The combination of a high technology sensor, together withadvanced signal conditioning and packaging techniques,
provides an ideal long term solution for reliable, accurate
and economical measurements.

• High over pressure capability

Bespoke as standard• Multiple electrical and pressure connector options

Custom-built from standard components, manufacturing
sensors to your requirement is fast and simple; each UNIK
5000 is a “bespoke” pressure sensing solution, but with theshort lead times and competitive pricing you would expectfrom standard products.

• Operating temperature ranges from –55 to +125°C

Expertise
We have the people and the knowledge to support yourneeds for accurate and reliable product performance;
our team of experts can help you make the right sensor• Hazardous area certifications• mV, mA, voltage and configurable voltage outputs(-67 to +257°F)• Online configuration tool

UNIK 5000 specifications
Measurement
Operating pressure ranges:
Gauge ranges
Any zero based range 70 mbar to 70 bar (1 to 1000 psi)

Supply and outputs
Electronics
optionDescription

Supplyvoltage (V)OutputCurrent
consumption(mA)

0

mV Passive

2.5 to 12

10 mV/V^<2 at 10 V

1mV Linearised

7 to 1210 mV/V^

<3

2

mA

7 to 28**

4-20 mA<30

30 to 5 V 4-wire

7 to 16**

0 to 5 V<3

4

0 to 5 V 3-wire

7 to 16**

0 to 5 V*

<3

5

Basic
Configurable(3-wire)

See
below

See
below

<3

6

0 to 10 V
4-wire12 to 16**

0 to 10 V<3

Wet/WetUni-directional or bi-directional 350 mbar to 35 bar (5 to 500 psi)

7

0.5 V to 4.5 V
Ratiometric

5.0 ± 0.5

0.5 to 4.5 V<3Line pressure: 70 bar (1000 psi) maximum

8

Configurable(4-wire)7 to 36

See
below

See
below

9

Configurable(3-wire)7 to 36

See
below

See
below

Note: All psi values are approximate.

Sealed gauge ranges
Any zero based range 10 to 700 bar (145 to 10000 psi)
Absolute ranges
Any zero based range 100 mbar to 700 bar (1.5 to 10000 psi)
Differential ranges
Wet/Dry
Uni-directional or bi-directional 70 mbar to 35 bar (1 to 500 psi)

Barometric ranges
Barometric ranges are available with a minimum span of 350
mbar (5.1 psi)
Non-zero based ranges
Non-zero based ranges are available. For non-zero based
gauge ranges, please contact Druck to discuss yourrequirements.
Over pressure• 10 × FS for ranges up to 150 mbar (2 psi)
• 4 × FS for ranges <350 mbar (5 psi)
• 6 × FS for ranges up to 700 mbar (10 psi)
• 2 × FS for barometric ranges• 4 × FS for all other ranges (up to 200 bar for ranges ≤70 bar
and up to 1200 bar for ranges >70 bar)*For differential versions the negative side must not exceed
the positive side by more than:• 6 × FS for ranges up to 150 mbar (2 psi)
• 4 × FS for ranges up to 700 mbar (10 psi)
• 2 × FS for all other ranges up to a maximum of 15 bar (200 psi)
Containment pressure• Ranges up to 150 mbar (2 psi) gauge 10 x FS
Ranges up to 70 bar (1000 psi) gauge 6 x FS
(200 bar (2900 psi) max)
• Ranges up to 70 bar (1000 psi) absolute 200 bar (2900 psi)
• Ranges above 70 bar (1000 psi)
• 1200 bar (17400 psi)*
Differential (-ve port) must not exceed positive port by morethan 6 × FS (15 bar (200 psi) maximum.
*600 bar (8700 psi) for pressure connector options PX, RA and RF^ with a 10 V supply mV output sensors give 100 mV over the full scalepressure.
• Output is ratiometric to the supply voltage• Output reduces pro-rata for pressure ranges below 350 mbar (5 psi)
*0 to 5 V 3-wire output is non true zero. At pressures below 1% of span the
output will be fixed at approximately 50 mV**32 V in non-hazardous area operation~ Supply voltage is between [Maximum output + 1 V] (7 V minimum) to 16 V
(32 V in non-hazardous area operation)

Basic configurable (option 5), configurable 4-wire (option8), configurable 3-wire (option 9)
Any pressure signal output configurations will be available,
subject to the following limitations:
Output specification

Basic configurable(option 5)

Configurable(options 8, 9)

Minimum span:

4V

2V

Maximum span:

10 V

20 V

Maximum output limit:

11 V

±10 V

Maximum zero offset:

Span / 2±Span

Current consumption:< 3 mA< 20 mA @ 7 Vdc
decreasing
to < 5 mA @ 32 Vdc

No

Yes

+125˚C+80˚C

Reverse output
response:
Maximum operating
temperature:

Output voltage range can be specified to a resolution of 0.1 V.
The output will continue to respond to 110% FS. i.e. if a 0 to 10 V output isspecified, the output will continue to increase proportionally to appliedpressure until at least 11 V.
Option 5: Not true zero, the output will saturate at < 50 mV.
Options 8, 9: On startup <100 mA drawn for 10 ms typically.
Options 8, 9: Shunt calibration: not available with reverse output.

mV passive

ExamplesConfiguration

AllowedNot Allowed

Basic Configurable(Option 5)

0 to 5 V

1 to 4 V (span too small)

0.5 to 4.5 V

4 to 11 V (offset too big)

1 to 6 V
1 to 11 V
Configurable(Options 8, 9)≤ 70 bar
Industrial/Improved: ±0.25% FS BSL
Premium not available> 70 bar
Industrial/Improved: ±0.5% FS BSL
Premium not available-10 to 0 V

0 to 12 V (outside ±10 V
limits)

Note 1: For the barometric pressure range, accuracy is of span, not fullscale.

0 to 5 V

6 to 10 V (offset too big)

-5 to 5 V

0 to 0.5 V (span too
small)

Note 2: For bi-directional ranges, accuracy is specified for each direction
separately.

-2 to 10 V
1 to 6 V
10 to 0 V

Power-up time
• mV, Voltage and current versions: 10 ms

Zero offset and span setting
Demountable electrical connector options allow access topotentiometers that give at least ±5% FS adjustment (seeelectrical connector section).
Factory set to:
Product Description

Industrial

Improved andPremium

Current and voltage versions(Demountable electrical connections
and cable gland)±0.5% FS

±0.2% FS

• 500 Vac: < 5 mA leakage current (mV and mA versions only)

Current and voltage versions (All otherelectrical connections)±1.0% FS

±1.0% FS

Shunt calibrationmV versions±3.0 mV±3.0 mV• Configurable 3-wire and 4-wire versions: 500 ms
Insulation• 500 Vdc: 100 MΩ

Shunt calibration provides a customer accessible connection
which, when applied, causes a shift in output of 80% FS inorder to simulate applied pressure. It is fitted to the mV,
configurable 4-wire and configurable 3-wire versions as
standard. It is not available with DIN, M12 x 1 or M20 x 1.5
electrical connectors (options 7, D, G and R).

Long term stability

Shunt calibration is activated in different ways depending on
the electrical connector and version:

Industrial accuracy performance:• mV versions: connect shunt cal to -ve supply or whereavailable, connect both shunt cal connections together.

• -10 to +50°C (+14 to +122°F):±0.75% FS TEB*

• -20 to +80°C (-4 to +176°F):±1.5% FS TEB• -40 to +80°C (-40 to +176°F):±2.25% FS TEB• -40 to +125°C (-40 to +257°F):±2.25% FS TEB• Configurable 4-wire and configurable 3-wire versions:connect Shunt cal to -ve output or, where available,
connect both shunt cal connections together.
Note: Not available with reverse output.

Performance specifications

There are three grades of performance specification:
Industrial, Improved and Premium.

Accuracy
Voltage, current and mV linearized
Combined effects of non-linearity, hysteresis andrepeatability:• Industrial: ±0.2% FS BSL• Improved: ±0.1% FS BSL• Premium: ±0.04% FS BSL±0.05% FS typical (±0.1% FS maximum) per year increasing
pro-rata for pressure ranges below 350 mbar
Temperature effects
Four compensated temperature ranges can be chosen.

*TEB = Temperature Error Band

Improved and premium accuracy performance:• -10 to +50°C (+14 to +122°F):±0.5% FS TEB• -20 to +80°C (-4 to +176°F):±1.0% FS TEB• -40 to +80°C (-40 to +176°F):±1.5% FS TEB• -40 to +125°C (-40 to +257°F):±1.5% FS TEB

Temperature effects increase pro-rata for pressure ranges
below 350 mbar (5 psi) and are doubled for barometricranges.
Line pressure effects (differential version only)
Zero shift: <±0.03% span/bar of line pressure
Span shift: <±0.03% span/bar of line pressure
Effects increase pro-rata for differential pressure ranges
below 700 mbar (10 psi).

Physical specifications
Environmental protection• See electrical connector section• Hyperbaric pressure: 20 bar (300 psi) maximum
Operating temperature range
See Electrical Connector section
Pressure media
Fluids compatible with stainless steel 316L and Hastelloy
C276.
For the wet/dry differential version, negative pressure port:
Fluids compatible with stainless steel 316L, stainless steel 304,
Pyrex, silicon and structural adhesive.
Enclosure materials
Stainless steel (body), nitrile- or silicone-rubber (o-rings,
gaskets), EPDM (gaskets), PVDF (depth cone), PTFE (ventfilter), nickel plated brass (lock rings), glass filled nylon(electrical connector assemblies), delrin (depth cone). Cable
sheaths as specified (see Electrical Connector).
Pressure connector
Available options are:• G1/4 female*• M20 X 1.5 male (8 mm bore)• Depth cone (G1/4 female open face)• M12 x 1.0 74° external cone
• Quick release male
• VCR female*• VCR male*• NW16 flange• R3/8 male
• R1/4 male
Choose connectors marked * for pressure ranges over70 bar. Other pressure connectors may be available, contact Druck to
discuss your requirement.

General certifications
CRN Certified 0F13650.517890YTN ADD1/REV1, 0F13828.2(sensor types K and O) and CSA 0F13650.56 ADD1 for pressure
ranges up to and including 350 bar (5000 psi).
Electrical connector
Various electrical connector options are available offering
different features:
Code
No.

-55 to +125

-67 to +257

-

Y

1

Cable Gland-40 to +80-40 to +176

65

N

2

Raychem Cable-55 to +125

-67 to +257

65

N

3

Polyurethane
Depth-40 to +80-40 to +176

68

N

4

Hytrel Depth-40 to +80-40 to +176

68

N

6/E

BayonetMIL-C-26482-55 to +125

-67 to +257

67

N

7

DIN 43650 FormA Demountable-40 to +80-40 to +176

65

Y

A/F

BayonetMIL-C-26482
Demountable-55 to +125

-67 to +257

65

Y

C

1/2 NPT Conduit-40 to +80-40 to +176

65

N

D

Micro DIN (9.4mm pitch)-40 to +80-40 to +176

65

N

G

M12x1 4pin

-55 to +125

-67 to +257

67

N

K

Zero Halogen
Cable
Demountable-40 to +80-40 to +176

65

Y

N

Kynar Depth-40 to +80-40 to +176

68

N

R

M20 x 1.5 Inline-40 to +80-40 to +176

65

Y

• G1/4 male flat long
• G1/4 male flat with snubber• G1/8 male 60° internal cone
• G1/2 male via adaptor*• 1/4 NPT female*• 1/4 NPT male• 1/8 NPT male• 1/8-27 NPT female• 1/2 NPT male via adaptor• 7/16-20 UNJF female*• 7/16-20 UNF male short flat*• 7/16 UNF long 37° flare tip*
• 7/16-20 UNJF male 74° external cone• 3/8-24 UNJF
• 1/4 swagelok bulkhead• M10 X 1 80° internal cone
• M12 X 1 60° internal cone
• M14 X 1.5 60° internal cone
• M20 X 1.5 male (3 mm bore)

Zero/
span
adjustNo connector0• G1/4 quick connect

IP rating°F• G1/4 male 60° internal cone

• G1/4 male with nipple

Maximum operating
temperature range°C• G1/4 male flat

• G1/4 male flat with cross bore protection

Description

Note 1: Electronics output options 8 and 9 are restricted to a maximum
operating temperature of 80°C (176°F).
Note 2: Hazardous area approved versions are restricted to a maximum
operating temperature range of -40°C to +80°C (-40°F to +176°F).
Note 3: Electrical connector option R IP65 rating only with suitable conduit/
cable fitting.

CE conformity

Hazardous area approvals (optional)• RoHS 2011/65/EU

General applications:• Pressure equipment directive 2014/68/EU soundengineering practice• IECEx/ATEX intrinsically safe ‘ia’ group IIC• ATEX 2014/34/EU (Optional)• EMC directive 2014/30/EU• BS EN 61000-6-1: 2007: Susceptibility - light industrial• BS EN 61000-6-2: 2005: Susceptibility - heavy industrial*• BS EN 61000-6-3: 2007+A1:2011: Emissions - light industrial• BS EN 61000-6-4: 2007+A1:2011: Emissions - heavy industrial• BS EN 61326-1: 2013: Electrical equipment for measurement,
control and laboratory use• BS EN 61326-2-3: 2013: Particular requirements for pressuretransducers*PMP/PTX models with electrical connector options 6, A, C, E, F, G, N and R
with cable screen connected to case.

• INMETRO intrinsically safe ‘ia’ group IIC• NEPSI intrinsically safe ‘ia’ group IIC• FM approved (Canada & US) intrinsically safe Exia Class I,
Division 1, Groups A, B, C & D and Class I, Zone 0 AEx/Ex ia
Group IIC; Single Seal
Mining applications:• IECEx/ATEX intrinsically safe ‘ia’ group I• INMETRO intrinsically safe ‘ia’ group I
For full certification details, refer to the type-examination certificates (orapproval listings) and supplied hazardous area installation instructions.

Electrical connector
Electronics option
Connector type

Molex

Cable
(Not Raychem)

Raychem Cable

BayonetOption code

4 to 20 mA

Voltage (3wire) and basic
configurable

Voltage (4-wire)

Configurable
voltage (4-wire)

Configurable
voltage (3-wire)

1 Red+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply

0

1, 3, 4,
C, N

2

6, A

mV

+ve Supply

2 Yellow-+ve Output+ve Output+ve Output+ve Output+ve Output

3 Green---ve Output-ve Output

0V Common-ve Output

4 Blue-ve Supply

0V Common-ve Supply-ve Supply

0V Common-ve Supply

5 Orange---

Shunt Cal

Shunt Cal

Shunt Cal

6 Black

Case

Case

Case

Case

Case

-

Red

+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply

Yellow-+ve Output+ve Output+ve Output+ve Output+ve Output

Blue

-

-

-ve Output-ve Output

0V Common-ve Output

White-ve Supply

0V Common-ve Supply-ve Supply

0V Common-ve Supply

Orange---Shunt Cal

Shunt Cal

Shunt Cal

Black

-

-

-

-

-

-

Screen

-

-

-

-

-

-

Red

+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply

White

-

+ve Output+ve Output+ve Output+ve Output+ve Output

Green

-

-

-ve Output-ve Output

0V Common-ve Output

Blue-ve Supply

0V Common-ve Supply-ve Supply

0V Common-ve Supply

Black

-

-

-

Shunt Cal

Shunt Cal

Shunt Cal

Screen

-

-

-

-

-

-

A

+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve SupplyB-ve Supply+ve Output+ve Output+ve Output+ve Output+ve Output

C

-

-

-ve Output-ve Output

0V Common-ve Output

D-

0V Common-ve Supply-ve Supply

0V Common-ve Supply

E

-

-

-

Shunt Cal

Shunt Cal

Shunt Cal

F

-

-

-

-

-

Shunt Cal

DIN A

7

1

+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply

Micro DIND

2

-ve Supply

0V Common-ve Supply-ve Supply

0V Common-ve Supply

BayonetE, F

Alternative Wiring
Options

M12 X 1G

4-Pin

Zero Halogen
Cable(Demountable)

M20 x 1.5 Female
Demountable3-+ve Output+ve Output+ve Output+ve Output+ve Output

E

Case

Case

-ve Output-ve Output

0V Common-ve Output

A+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply

B-

0V Common-ve Supply-ve Supply

0V Common-ve SupplyC-+ve Output+ve Output+ve Output+ve Output+ve Output

D-ve Supply--ve Output-ve Output

0V Common-ve Output

E

-

-

-

Shunt Cal

Shunt Cal

Shunt Cal

F

-

-

-

Shunt Cal

Shunt Cal-1

+ve Supply

2

K

R

+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve Output+ve Output+ve Output+ve Output+ve Output

3-ve Supply

0V Common-ve Supply-ve Supply

0V Common-ve Supply

4

Case

Case

-ve Output-ve Output

0V Common-ve Output

Pink+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply+ve Supply

White

-

+ve Output+ve Output+ve Output+ve Output+ve Output

Green

-

-

-ve Output-ve Output

0V Common-ve Output

Blue-ve Supply

0V Common-ve Supply-ve Supply

0V Common-ve Supply

Grey

-

-

-

Shunt Cal

Shunt Cal

Shunt Cal

Brown

-

-

-

-

-

-

Yellow

-

-

-

-

-

-

Screen

-

-

-

-

-

-

+ve

+ve Supply

-

-

-

-

-

-ve

-ve Supply

-

-

-

-

-

Ordering information (see the online configuration tool at Druck.com)(1) Select model number

Main Product Variant
PMP
Amplified Pressure Transducer
PDCR
mV Pressure Transducer
PTX
4-20 mA Pressure Transmitter
Product Series
5
UNIK 5000
		
Diameter and Material
		025mm Stainless Steel			
Electrical Connector (Note 6)
			0
No Electrical Connector (Note 7)
			1
Cable Gland (Polyurethane Cable)
			2
Raychem Cable
			3
Polyurethane Cable (Depth)
			4Hytrel Cable (Depth)
			6
MIL-C-26482 (6-pin Shell Size 10) (Mating connector not supplied)			7
DIN 43650 Form A Demountable (Mating connector supplied)			A
Demountable MIL-C-26482 (6-pin Shell Size 10) (Mating connector not supplied)			C
1/2” NPT Conduit (Polyurethane cable)
			D
Micro DIN (9.4 mm Pitch) (Mating connector supplied)
			E
MIL-C-26482 (6 pin Shell Size 10) Alternative Wiring (Mating connector not supplied)			F
Demountable MIL-C-26482 (6 pin Shell Size 10) Alternative Wiring (Mating connector not supplied)			G
M12 x 1 4-pin male (Mating connector not supplied)			K
Zero Halogen Cable Demountable
			N
Kynar Cable (Depth)
			R
M20 x 1.5 Inline Female Conduit Demountable (Note 8)
				Electronics Option
				0
mV Passive 4-wire (PDCR) (Note 1)
				1
mV Linearised 4-wire (PDCR)
				2
4 to 20 mA 2-wire (PTX)				30 to 5 V 4-wire (PMP)				4
0 to 5 V 3-wire (PMP)				5
Basic Configurable 3-wire (PMP)				6
0 to 10 V 4-wire (PMP)				7
0.5 to 4.5 V Ratiometric 3-wire (PMP) (Note 5)
				8
Configurable 4-wire (PMP) (Note 4, 5)
				9Configurable 3-wire (PMP) (Note 4, 5)					Compensated Temperature Range					TA-10 to +50 °C (14 to +122 °F)					TB-20 to +80 °C (-4 to +176 °F)					TC-40 to +80 °C (-40 to +176 °F)					TD-40 to +125 °C (-40 to +257 °F) (Note 2, 5)						Accuracy						A1
Industrial						A2
Improved						A3
Premium							Calibration							CAZero/Span Data							CBRoom Temperature							CCFull Thermal							CRRoom Temperature/Measured Reading								Hazardous Area Approval (Note 6)								H0None								H1IECEx/ATEX Intrinsically Safe ‘ia’ Group IIC
								H2IECEx/ATEX Intrinsically Safe ‘ia’ Group I								H6FM (C & US) Intrinsically Safe ‘ia’ Group IIC/ABCD								HAIECEx/ATEX Intrinsically Safe ‘ia’ Groups I/IIC [H1 + H2]
								HS	IECEx/ATEX/FM (C & US) Intrinsically Safe ‘ia’ Groups 		
IIC/ABCD [H1 + H6]
								J1IECEx/ATEX/NEPSI Intrinsically Safe ‘ia’ Group IIC
								JAINMETRO Intrinsically Safe ‘ia’ Group IIC
								JBINMETRO Intrinsically Safe ‘ia’ Group I								JFINMETRO Intrinsically Safe ‘ia’ Group I/IIC [JA + JB]
									Pressure Connector									PAG1/4 Female (Note 3)														PBG1/4 Male Flat														PCG1/4 Male 60° Internal Cone									PDG1/8 Male 60° Internal Cone									PE1/4 NPT Female (Note 3)				
									PF1/4 NPT Male														PG1/8 NPT Male									PHM20x1.5 Male (3 mm bore)									PJM14x1.5 60° Internal Cone													PKM12x1 Internal Cone														PL7/16-20 UNJF Male 74° External Cone												PNG1/2 Male via Adaptor (Note 3) 				G1/4 Quick Connect														PQ									PR1/2 NPT Male via Adaptor (Note 3)			
									PS1/4 Swagelok Bulkhead													PTG1/4 Male Flat Long														PU7/16-20 UNF Long 37° Flare Tip (Note 3)			
									PV7/16-20 UNJF Female (Note 3)									PWDepth Cone (G1/4 Female Open Face)								PX7/16-20 UNF Male Short Flat (Note 3, 9)									PY3/8-24 UNJF
									PZM10 x 1 80° Internal Cone									RAVCR Female (Note 3, 9)				
									RBG1/4 Male Flat with Snubber									RCG1/4 Male Flat with Cross Bore Protection									RDM12 x 1.0 74° External Cone									REQuick Release Mount									RFVCR Male (Note 3, 9)									RJM20 x 1.5 Male (8 mm bore)									RQNW16 Flange									RT1/8-27 NPT Female									RUR3/8 Male									RVR1/4 Male									RWG1/4 Male with Nipple
PTX

5

0

7

2

-

TA

-

A2

-

CB

-

H0

-

PA

(Example Model Number)Ordering notes

4) State cable lengths and units

Note 1: Premium accuracy is not available on this version

Integer values only, e.g. 1 m cable, 8 ft. Minimum length 1 m (3ft) cable (only required with certain electrical connectors).

Note 2: Please ensure that the electrical connector selected is option 0, 2,
6, A, E, F or G.
Note 3: Select one of these pressure connectors for pressure ranges over70 bar
Note 4: Maximum operating temperature is 80°C (176°F)

5) Output options 5, 8 and 9

Note 5: Hazardous area certifications not available
Note 6: Hazardous area certifications are restricted by electrical connector
options in line with the following table:

Approval

0

1

2

3

4

6/E

7

A/F

C

D

G

K

N

R

H0

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

H2

Y

-

Y

Y

Y

Y

-

Y

-

Y

-

-

Y

H6

HA

Y

Y

Y

Y

Y

Y

Y

HS
J1

Y

JA
JB

Y

JF

Y

Y

-

Y
-

-

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y
-

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

-

Y
-

-

-

Y
-

-

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

Y

-

Y
-

-

Y

Y

Y

Y

Y

Y

Y

Y

-

-

-

-

-

-

-

-

Note 8: Electronics option 2 only
Note 9: Pressure ranges less than 500 bar2) S tate pressure range and units(e.g., 0 to 10 bar, -5 to +5 bar)Pressure unit options are:Symbol		Description
bar		
bar
mbar 		millibar
psi		pounds/sq. inch
Pa		PascalhPa		hectoPascalkPa		kiloPascal
MPa		MegaPascal
mm water
mmH2O		cm water
cmH2O		metres water
mH2O		
inches water
inH2O		feet water
ftH2O		mmHg		mm mercury
inHg		inches mercury
kg force/sq. cm
kgf/cm2		
atm		atmosphere
Torr		torr
3) State pressure reference (e.g., gauge)• gauge• absolute• barometric• sealed gauge• wet/dry differential• wet/wet differential

PTX5012-TB-A2-CA-H0-PA: 0 to 10 bar, gauge, 3 m cableY

PMP5028-TC-A3-CC-H0-PE: -15 to 75 psi, gauge, 15 ft cable,
output voltage -1 to 5 volts-Y

PDCR5071-TB-A1-CB-H0-PA: 0 to 100 bar, sealed gauge

-

Y

-

Y

-

-

-

-

-

Note 7: Available with component certification, use of which requires
incorporation into certified apparatus with an IP rated enclosure
appropriate to the certification type supplied

Reference options are:State voltage output at minimum and maximum pressure(e.g., output –1 to 9 V)
Typical order examples

Connector

H1

Maximum cable length 100 m (300 ft) for approval options
not H0; 200 m (600 ft) for approval option H0.

-

-

Y
-

-

Accessories (order as separate line items)
P/N 202-034-05: Sensor termination enclosure
P/N 600-914: Desiccant gel pack for STE
P/N 222-117-01: Short sink weight (to fit G1/4 female)P/N 222-179-01: Depth cone injection moulded for PW
connection
P/N 163-009: Mating connector for MIL-C-26482 (Electricalconnector options 6, A, E, and F) (Note 10)

Cable assemblies (Note 10)

A made up electrical connector with a length of cable.
1) Select part number
Main product (Note 10)
UNIKCABLE
Cable assembly		Electrical connector
		6
MIL-C-26482
		7
DIN 43650
		D
MICRO DIN (9.4 mm Pitch)
			Cable			1	Polyurethane cable (Note 11)			2Raychem cable

UNIKCABLE(Example part number)

-

6

-

2

2) State cable length and units (Integer value only)
Minimum length 1 m (3 ft)
Maximum length 200 m (600 ft)
Example: UNIKCABLE-6-2 5 m
Note 10: Not considered suitable for use in hazardous areas.
Note 11: This cable is the same as that used in the electrical connector
cable gland (Option 1 in the main product).

Mechanical drawings
7(0.27)

25 TYP (0.98)

72(2.83)

15(0.53)

Ø25 (0.98)

Ø25 (0.98)
UNIK 5
PRESSU#### 50##
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. GR

MALE PRESSURE CONNECTION

4CABLE GLAND WITH POLYURETHANE CABLE

HIGH PRESSURE
CONSTRUCTION( 22 A/F)(0.86)

21 TYP (0.82)

65 (2.56)

7(0.27)

15.5 (0.61)

22 A/F(0.86)UNIK 5
PRESSU#### 50##-T*********** AC
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. GR

Ø25 (0.98)

Ø 25 (0.98)

RAYCHEM CABLE

MEDIUM PRESSURE
CONSTRUCTION

16(0.62)

60(2.36)

69.5(2.73)

15.5(0.61)

DEPTH CABLE

Ø25 (0.98)
UNIK 5
PRESSU

25 TYP(0.98)#### 50##
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. GR

DEPTH CONE
PRESSURE ADAPTOR

Ø25 (0.98)

17(0.66)

LOW/MEDIUM PRESSURE
CONSTRUCTION

4(0.15)

15.5 BOTH ENDS
(0.61)

Ø25 (0.98)

Ø25(0.98)OPTIONAL
PRESSURE ADAPTORBAYONET MIL-C-26482
NON-DEMOUNTABLE

Ø

49(1.92)

Ø25 (0.98)

UNIK 5
PRESSU#### 50##-T*********** AC
S/N ********** TO *** baSupply: *** T
Output: *** T
Temp. Range:

DRUCK LTD. GR

72(2.83)

WET/WET & WET/DRY
DIFFERENTIALCONSTRUCTION
49 (1.32)
80(3.15)

28 (1.10)

35 (1.37)

NOTES:[1] DIMENSIONS SHOWN ARE FOR STANDARD LENGTH PRODUCTS WITH THE FOLLOWINGELECTRICAL OUTPUT OPTIONS:
mV LINEARISED (PDCR)
4 TO 20 mA (PTX)
STANDARD VOLTAGE OPTIONS (PMP)
BASIC CONFIGURABLE (PMP)
FOR mV PASSIVE (PDCR) - SUBTRACT 10 mm (0.39 in)
FOR CONFIGURABLE 4-WIRE AND CONFIGURABLE 3-WIRE (PMP) - ADD 15 mm (0.59 in)Ø30 (1.18)

42(1.65)[2] REFER TO PAGE 4 FOR LIST OF PRESSURE CONNECTION OPTIONS(ORIENTATION NOT CRITICAL)

32(1.25)[3] ALL DIMENSIONS ARE IN MILLIMETRES (INCHES IN PARENTHESES)[4] HIGH PRESSURE IS >70 BARMEDIUM PRESSURE
INDUSTRIAL ACCURACY >1 BAR TO < 50 BAR
LOW/MEDIUM PRESSURE
INDUSTRIAL ACCURACY < 1 BAR, > 50 BAR TO < 70 BAR
IMPROVED/PREMIUM ACCURACY < 70 BAR

29(1.14)
16(0.62)DIN 43650 DEMOUNTABLE

BAYONET MIL-C-26482
DEMOUNTABLE

15.5 (0.61)

Ø25 (0.98)
Ø25 (0.98)

CONDUIT WITH POLYURETHANE CABLE(22 A/F)

M12x1 4-PIN

47 (1.85)

39 (1.5)

16.25 (0.64)

33.45 (1.3)
Ø25 (0.98)

Ø 25 (0.98)

Ø 44 (1.73)

15.5 (0.6)
MICRO DIN (9.4 mm)

M20 x 1.5 INLINE FEMALECONDUIT DEMOUNTABLE

Ø30 (1.18)

ZERO HALOGEN CABLE DEMOUNTABLE

Copyright 2021 Baker Hughes Company. All rights reserved.

920-483M

BHCS38660		

(06/2021)

druck.com