Ranger Pro Sensor Interface
Datasheet

Bently Nevada Machinery Condition Monitoring

165M4206 Rev. B

Description
The Ranger Pro Sensor Interface and external sensor
combination is a smart wireless device that is engineered forquick installation and setup to monitor machine health andcompliments existing Ranger Pro solutions (datasheet
125M5237). An applicable sensor is connected to the Ranger
Pro Interface module by a short cable. Using industry
standard wireless protocols such as ISA 100.11a andWirelessHART, the device complies with international,
industrial wireless networking standards engineered to serve
the needs of process industries.
The Ranger Pro Sensor Interface enables one applicablesensor to connect via cable to a single Ranger Pro interface
module (two sensor connectivity will be available in a future
release). This system measures overall velocity, acceleration,with spectrums/waveforms plus temperature ranging from
low to high speed applications. Other measurement
modalities are forthcoming.
The Bently Nevada Ranger Pro Sensor Interface enables you
to:
l

l
l
l
l
l
l

Monitor and optimize the reliability of low- andmedium-criticality machines.
Establish or expand existing reliability programs.
Make maintenance decisions based on current data.
Reduce maintenance costs.
Decrease unplanned machine failures.
Increase machinery life.
Multi-channel support

Ranger Pro Sensor Interface is a simple, easy to implement
solution for use in hazardous or difficult to accessRanger Pro Sensor Interface
Datasheet

165M4206 Rev. B

environments such as elevated temperatures
and tight or enclosed installations.
Use the Ranger Pro Sensor Interface to get
short- and long-term trending data, anddiagnostic reporting.
Quickly publish overall data through Modbus tothird-party tools or spectra and waveform data
through Generic Client or Hart IP Interface toBently Nevada System 1 software. Configure
Ranger Pro devices over-the-air using thirdparty tools or the Ranger Pro Configuration
software.

2/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

System 1 Support

Machinery Applications

Ranger Pro collects overall vibration,
temperature measurements, timebase
waveforms, spectra, and PeakDemod spectrum
using Generic Client Interface (GCI) for ISA100
Ranger Pro devices and HART IP for
WirelessHART Ranger Pro devices with System 1
software. You can filter overall and dynamic
timebase and spectra data.

Ranger Pro Sensor Interface is a vibration
sensor for machines with roller-element
bearings including:
l
l
l
l
l
l
l
l
l

Agitators
Air compressors
Ball mills
Blowers
Centrifuges
Cooling tower fans and pumps
MotorsSmall reciprocating compressors
Small hydro and steam turbines

Network Installation
A typical network installation uses several
Ranger Pro Sensor Interface sensors, Ranger
Pro repeaters, wireless device managers, andaccess points. Ranger Pro Interface Module isavailable in either WirelessHART or ISA100
configuration.

Hardware Features

You can use third-party tools or the Ranger Pro
Configuration software to quickly provision andconfigure Ranger Pro devices over-the-air.

You can configure Ranger Pro Sensor Interface
to work in a variety of environments andapplications.
l

l

l

l
l
l

l

Tri-axial capable velocity andacceleration detection.
Machine surface temperature
measurement.
Mounting hardware options to fit mostapplications and integral alignment
capability
Replaceable lithium-chloride battery.
IP67 dust and water resistant.
Embedded sensors connect using the
ISA100 wireless or WirelessHART network
protocols.
Can act as a router for other Ranger Pro
sensors.

Wireless range varies depending on
environmental obstacles, gateway antenna
type, and the orientation of the sensor relative
to the gateway antenna.

3/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Hazardous Area Approvals

Compliance and Certifications

WARNING

This device complies with part 15 of the FCC Rules.
Operation is subject to the following two conditions:

HAZARDOUS ENVIRONMENT

1. This device may not cause harmfulinterference.

Risk of explosive atmosphere.
Avoid electrostatic potential,
especially on plastic
components. Adapters,
isolation valves, or sealing rings
that are not compatible withprocess gasses will corrode andfail. This failure may result in
gas leaks, fire, explosion, or
projectiles.

2. This device must accept any interference
received, including interference that may
cause undesired operation.

EMC
conformity
standards

IEC 61326-1, ETSI EN 301 489-1,
CISPR22, ETSI EN 301 489-17

Radio
spectrumETSI EN 300 328

SafetyETSI EN 61010-1, IEC 62479

Hazardous
Atmosphere

CSA
Class I, Division1, Group A, B,
C, D T4
Class II, Division1, Group F, G
T135°
ATEX/IECEx Zone 0

Conformity

Compliant with all CE and
FCC/IC requirements

Validfor RangerPro
BN P/N

147M7136-01-11
147M7136-02-11
160M0016-01

De-energize all devices before
placement or removal. To
prevent corrosion and failure,verify that all components arecompatible. Verify thathazardous materials,
atmospheres, and conditions
have been removed or that
relevant risk mitigation
measures have been
implemented.

North
America(US and
Canada)CSA
Class I, Zone 0, AEx ia IIC T4 Ga
Zone 20, AEx iaD IIIB T135°C Da
Class I, Division1, Group A, B, C, D
T4Class II, Division1, Group F, G T135°

IECEx

Ex ia IIC T4 Ga
Ex ia I Ma
Ex ia IIIB T135°C Da

ATEX

4/28

II 1G Ex ia IIC T4 Ga
I M1 Ex ia I Ma
II 1D Ex ia IIIB T135°C Da

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Specifications
Ranger Pro Tethered Accelerometer 70M503
Feature
Axis

Characteristic
X, Y, Z

Sensing element
Transverse sensitivity(Typical)Transverse sensitivity(Typical)Resonant frequency
Total noise(acceleration rms)

Total noise(velocity rms)

Temperature< 7% (160 Hz)±5% (160 Hz)Z

39 mV/g

X, Y

35 mV/g

Amplitude linearity
Frequency response3
Micro-electromechanical sensors (MEMS)

Sensitivity tolerance
Sensitivity

Value

± 2%
Z (± 3 dB)

0.3 Hz to 10 kHz

X, Y (± 3 dB)

0.3 Hz to 2.5 kHz

Z> 15 kHz

X, Y

5 kHz

Z: 0.3 to 10 kHz

0.003 g

X, Y: 0.3 to 2 kHz

0.004 g

Z: 1-200 Hz

0.18 mm/s [0.007 in/s]

Z: 10-1000 Hz

0.09 mm/s [0.004 in/s]

X,Y: 1-200 Hz

0.57 mm/s [0.022 in/s]

X,Y: 10-1000 Hz

0.11 mm/s [0.004 in/s]

Sensor type

Digital Resistance Temperature Device (RTD)

Measurement range-40°C to 125°C (-40°F to 257°F)

Resolution

0.1°C (0.06°F)

Accuracy±1°C (typical), ±2°C (maximum)

Measurement Interval

10 min, 20 min, 30 min, 1 h, 2 h, 3 h, 4 h, 6 h

5/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

FeatureCharacteristic

Electrical

MechanicalEnvironmental

Value

Supply Current

2.09 mA (max)

OutputMultiplexed output

GroundingCase isolatedConnector

Top-exit, M12 A-coded, 5 pin

Material

316 Stainless

Mounting

Integral M20x1.0 Coupling Nut

Weight62 g

Sealing

Hermetic

Operating temperature-40°C to 125°C (-40°F to 257°F)

IP Rating

IP66/67 when mated to connector

Shock5000 gAltitude< 3000 m

Ranger Pro Sensor Interface: P/N 70M323 and 70M423
Trended Variables
Trend Variables apply when connected to a 70M503 Ranger Pro accelerometer.
Characteristic

Value

Temperature
Measurement range-40°C to 125°C (-40°F to 257°F)

Resolution

0.1°C (0.06°F)

Measurement Interval

10 min, 20 min, 30 min, 1 h, 2 h, 3 h, 4 h, 6 h

Acceleration
Acceleration amplitude range

0 – 200 m/s2 (0 - 20 g)

Acceleration units / subunits

g or m/s2 / peak or rms

Fmin

0.3, 2, 5, 10, 100, 200 (Hz)Fmax

200, 500, 1000, 2000, 5000, 10000‡Hz‡10,000 only on Z-axis

Measurement Interval

10 min, 20 min, 30 min, 1 h, 2 h, 3 h, 4 h, 6 h

6/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Characteristic

Value

Velocity
Velocity amplitude range

0 – 50 mm/s (0 - 2 in/s)

Velocity units / subunits

in/s or mm/s peak or rms

Fmin

1, 5, 10 Hz

Fmax

200, 500, 1000, 2000 Hz

Measurement Interval

10 min, 20 min, 30 min, 1 h, 2 h, 3 h, 4 h, 6 h

PeakDemod
PeakDemod Pk

Z axis only
Parameters based on PeakDemod Spectrum
settings below

Measurement interval

6 h, 8 h, 12 h, 1 d, 2 d, 7 d, 14 d, 28 d

Waveforms and Spectra
Characteristic

Value

Acceleration
Acceleration waveform

X, Y and Z axis

Fmin0.3, 2, 5, 10 Hz

Fmax

200, 500, 1000, 2000, 5000, 10000 ‡ Hz
‡ Z-axis only

Number of samples

1024, 2048, 4096, 8192

Units/subunits

g or m/s2 / peak

Measurement Interval

6 h, 8 h, 12 h, 1 d, 2 d, 7 d, 14 d, 28 d

Velocity
Velocity spectra

X, Y and Z axis depending on sensor model

Fmin

5, 10

Fmax

200, 500, 1000, 2000

Number of lines

400, 800, 1600, 3200

Units/subunits

in/s or mm/s / rms

Measurement Interval

6 h, 8 h, 12 h, 1 d, 2 d, 7 d, 14 d, 28 d

7/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Characteristic

Value

PeakDemod
PeakDemod spectrum or waveform

Z-Axis only

Fmax

200, 500, 1000, 2000, 5000 Hz

Demod Band Min

500, 1000, 2000, 5000 Hz

Units/subunits

g, m/s2 / peak

Measurement Interval

6 h, 8 h, 12 h, 1 d, 2 d, 7 d, 14 d, 28 d

Output Data
Characteristic

Value

Trended Variables
ISA100 Temperature

wHART Temperature

ISA100 Acceleration

wHART Acceleration

ISA100 Velocity

wHART Velocity

ISA100 PeakDemodwHART PeakDemod

Modbus: Supported
GCI: Supported
Modbus: Supported
HART-IP: Supported
Modbus: Not supported
GCI: Future Release
Modbus: Supported
HART-IP: Supported
Modbus: Supported
GCI: Supported
Modbus: Supported
HART-IP: Supported
Modbus: Not supported
GCI: Future Release
Modbus: Supported (Channel 1 only)
HART-IP: Supported

Waveforms and Spectra
Requires System 1 and either Generic Client Interface (GCI) or HART IP.

8/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Wireless
Characteristic

Value

Network standard

ISA100.11a, WirelessHART

Network topology

Star (ISA100) or mesh (ISA100 or WirelessHART)

Radio standard

IEEE 802.15.4

Radio frequency

2.45 GHz ISM band

Provisioning/ firmware updates

Over-the-air or via the USB docking station.

Encryption/ security

128-bit AES encrypted packets

Output power

5.13 dBm, typicalWireless range

150 meters sensor to access point, 100 meters
sensor to sensor, line of sight. (Actual range
depends on obstacles present, gateway antenna
type, and orientation of the sensor relative to thegateway antenna.)

147M7136-01 ISA100 Device
Typical Conducted Power

8.7 dBm (7.4 mW)

ModulationOQPSK, DSSSChannel BW

5 MHz,

Operating Frequency

2.405 to 2.48 GHz

147M7136-02 WirelessHART Device
Typical Conducted Power

5.1 dBm (3.3 mW)

ModulationOQPSK, DSSSChannel BW

5 MHz

Operating Frequency

2.405 to 2.475 GHz

9/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Battery and PowerCharacteristic

Value
Replaceable D size 3.6V lithium-thionyl chloride
with standard button-top termination.

Type

Warning: Use only one of the following
batteries: Tadiran TLH-5930, Tadiran TL-5930,
Tadiran SL-2780, or Xeno Energy XL-205F.

Up to five years depending on the operating
mode and configuration.

Life

Battery models
Hazardous area temperature range (Ta)Temperature range

TLH-5930-40°C < Ta < 80°C

TL-5930, Xeno XL-205F,
Tadiran SL-2780-40°C < Ta < 70°C

Environmental Conditions
Characteristic

Value

Operating temperature-40°C to 85°C (-40°F to 185°F) (Operating atextreme temperatures or beyond negatively
affects battery life and may damage the device.)

Vibration limit

20 g peak

Chemical resistance

Stainless steel and high temperature, solventand UV-resistant PPS plastic.

Shock resistance

0.5 meter drop onto concrete

Altitude

Maximum 3,000 m (9,842 ft.) outdoors

IP rating

IP67 dust and water resistantPhysical/Mechanical
Characteristic

Value

Weight

400 grams with battery

Dimensions

Height: 100 mm; diameter: 40 mm

Case material

316 stainless steel body and glass-reinforced,
impact-resistant PPS top

Mounting holeM6 x 1 mm X 6.5 mm deep internal thread

10/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Regulatory Compliance
Characteristic

Value

EMC conformity standards

IEC 61326-1, ETSI EN 301 489-1, CISPR22, ETSI EN 301
489-17

Radio spectrum

ETSI EN 300 328

SafetyETSI EN 61010-1, IEC 62479

Hazardous Atmosphere

CSA Class 1 Division 1 Groups A, B, C, D T4
ATEX/IECEx Zone 0
Class II, Division1, Group F, G T135°

Conformity

Compliant with all CE and FCC/IC requirements

Valid for Ranger Pro BN P/N

70M323, 70M423, 70M503

Entity Parameters
CharacteristicValue
Lo: 800 μH Co: 3.78 μF

Ranger Pro Interface Modules

Uo: 5.88V Io: 196 mA
Po: 288 mW
Lo: 0 μH Co: 1.87 μF

Ranger Pro Triaxial Accelerometer

Uo: 11.1V Io: 249 mA
Po: 450 mW

ISA100.11a compatible gateways †Characteristic
Bently Nevada

Value
Bently Nevada 70M320 ISA100.11a Gateway
Up to 50 Ranger Pro devices per Gateway
See the Ranger Pro Gateway Datasheet 157M8584
YFGW 410 Field Wireless Management Station
Up to 4 access points = 160 sensors

Yokogawa

YFGW 510 and YFGW 520 Field Wireless Access
Points. Up to 40 Ranger Pro sensors per accesspoint.

11/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Characteristic

Value
WDM Wireless Device Manager R310.2-4 or newer
Up to 8 access points = 320 sensors

Honeywell
FDAP Field Device Access Point
Up to 40 Ranger Pro sensors per access point
Ranger Pro sensor catalog number 70M323 isrecommended for ISA100a Gateway
Number of Hops
(Depth to Gateway)

3

WirelessHART compatible gateways.†
Characteristic

Value

Emerson 1410S

1410S (compatible with firmware version 6.4.5 ornewer) up to 200 Ranger Pro sensors per gateway

Emerson 1410A/B/D

1410 (compatible with firmware version 4.7.84 or
newer)
up to 70 Ranger Pro sensors per gateway

Emerson 1420

1420 (compatible with firmware version 4.7.84 or
newer)
up to 70 Ranger Pro sensors per gateway

Ranger Pro sensor catalog number 70M423 isrecommended for WirelessHART Gateway
Number of Hops
(Depth to Gateway)

3† Generic Client Interface (GCI) or HART IP required. Order when new or license as necessary.

Accelerometer to Interface Module Cable
Characteristic

Value

Mechanical
Length5 m, 10 m

Number of Conductors

5Connector Type

M12 A-Code 5-Pin Plug and socket, straight

Connector Contact Pin

Gold Plated Copper Alloy

12/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Characteristic

Value

Connector Housings

Stainless Steel

Cable Jacket

Black XLPE

Cable Shield

Braided (≥ 90% coverage), Tinned-Copper.
Electrically terminated to Interface moduleconnector only.

Cable Diameter0.3 in, typical

Bend Radius

Diameter x 12

Electical
Rated Voltage

300 Vdc

Environmental
Operating Temperature-40°C to 125°C

IP Rating

IP67 when mated

RoHS

Compliant

REACH

Compliant

UV ResistanceYes

Flamability

FT2

System 1
v21.1 or higher. Refer to System 1 121M7997 release notes for compatibility guidelines.

Advanced Features
Characteristic

Value

Data on Demand
Mode

User-initiated. Acquisition initiated from Ranger
Pro Configuration Software.

StatusIdle, Requested or Busy

Data on Vibration
ThresholdUser settings. Range: 0 to 0.1 in/s rms

Mode

Enabled/Disabled

Detection

XYZ vector sum or Z axis only

13/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Characteristic

Value

StatusOn or Off

Data on Severity

Vibration

Mode

Enabled/Disabled

TA Proven Method Level 3

User settings. Range: 0.05 to 2.5 in/s rms

TA Proven Method Level 4User settings. Range: 0.05 to 2.5 in/s rms

Detection

XYZ vector sum or Z axis only

StatusGreen, Yellow or Red when enabled

Data on Severity

Temperature

Mode

Enabled/Disabled

User Setting

32 to 257°F
0 to 125°C

Detection

Temperature of sensor baseStatusGreen, Yellow, or Red when enabled

14/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Ordering Information
Ranger Pro Acceleration
Interface

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

WirelessHART 70M423-AA-BB

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

ISA100 70M323-AA-BB
Ranger Pro versions 70M323, 70M423
use the same ordering information.

Ranger Pro Accelerometer
70M503-AA-BB-CC-DD

A: Battery Option

A: Mounting Options
00
No Stud01
M20x1.0 to 1/4-28
02
M20x1.0 to 3/8-24
03
M20x1.0 to 10-32
04
M20x1.0 to M6x1.005
M20x1.0 to M8x1.25
06
M20x1.0 to Cementing pad
07
M20x1.0 to 0.25x1.0 in Motor fin mount08
M20x1.0 to 0.25x1.75 in Motor fin mount09
M20x1.0 to 0.5x1.25 in Motor fin mount10
M20x1.0 to 0.5x2.0 in Motor fin mount
B: Connector Type0100

No battery

01

Battery supplied, not installed

B: Agency Approval Option
01

North America Cl1 Div1

02

ATEX/IECEx

Ranger Pro Installation Kit
130M5452 – AA
Description: Installation kit including battery
installation tool, O-rings, wrenches, andUSB readers.

Top Exit

C: Cable00
No Cable05
5 meter
10
10 meter
D: Approvals: Agency Approval Option
01
North America Cl1 Div1
02
ATEX/IECEx

A: Installation Package

15/28

00

Installation Tools

01

Installation Tools and USB Reader

02

USB Reader only

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

System 1 Ranger Pro Device License
3071/13 – AA-BB-CC
Description: System 1 device license for Ranger
Pro installed for use with System 1. One device
license is required per interface module.
A: Not Applicable for Ranger Pro
00
B: Not Applicable for Ranger Pro
00
C: Ranger Pro Device00## Number of licenses required
Option 3071/13 is only applicable toRanger Pro devices that are installed
for use with System 1. To order System
1, see System 1 Software Package
Datasheet (document 108M5214). The
AA option is only for vbOnline Pro
device licenses. The BB option is only
for 2300 monitor device licenses.

16/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Spare Mounting Adapters
Illustrations shown are not to scale. All mounting adapters are made from 316 stainless steel.
Units are mm[in]Part Number

Size

Illustration

Standard Studs

Applies to all standard studs.

164M6491

M20x1.0 to 1/428 UNF

164M6495

M20x1.0 to 3/8- 24 UNF

17/28

Ranger Pro Sensor Interface
DatasheetPart Number

Size

164M6487

M20x1.0 to#10-32

164M6489

M20x1.0 toM6x1.0

164M6493

165M4206 Rev. B

Illustration

M20x1.0 toM8x1.25

18/28

Ranger Pro Sensor Interface
DatasheetPart Number

165M4206 Rev. B

Size

Illustration

Universal Magnetic Mounting Adapter
02200371

1.85” Ø x 1.09” H (47 x 27.7 mm), 100 lbf (45kg) pull, 2-pole, ¼-28 female UNF thread.
Requires mounting option -01.

Cementing Pads and Adhesive

164M6490

M20x1.0 to
epoxy
cementingpad167236-01

3.5 g Click Bond CB200 acrylic adhesive for use with epoxy cementing pads. Sufficient
for about four pads.

19/28

Ranger Pro Sensor Interface
DatasheetPart Number

165M4206 Rev. B

Size

Illustration

Motor Fin Adhesive Mounts

164M6492

M20x1.0 to
0.25" x 1.0"

20/28

Ranger Pro Sensor Interface
DatasheetPart Number

164M6486

165M4206 Rev. B

Size

Illustration

M20x1.0 to
0.25" x 1.75"

21/28

Ranger Pro Sensor Interface
DatasheetPart Number

164M6488

165M4206 Rev. B

Size

Illustration

M20x1.0 to 0.5"
x 1.25"22/28

Ranger Pro Sensor Interface
DatasheetPart Number

164M6485

165M4206 Rev. B

Size

Illustration

M20x1.0 to 0.5"
x 2.00"

Expect a decrease in X, Y, Z accuracy when using fin type mounts.

23/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Accessories
The installation kit (1305452) includes a battery installation tool, two installation wrenches, spare Orings, and USB docking station. These parts can also be ordered individually.
Product orDocument

Item

164M6494

Ranger Pro Interface Module Mounting
Bracket and fasteners

159M7787

Ranger Pro Cap: additional protection for highmoisture installations

121M7993

Battery installation tool

160M0017

5 meter cable

160M0018

10 meter cable

129M0166

Sony USB configuration docking station

146M4035

Case O-ring 35 x 1 mm (qty. 20)

146M4036

E-module O-ring 34 x 1 mm (qty. 20)

125M3923

D-sized lithium-thionyl chloride 3.6 V battery

121M7997

Ranger Pro Sensor Interface
configuration software (not available for order, it is available for no charge from
Bently Nevada Technical Support)

125M6113

Ranger Pro Sensor Interface User Guide

24/28

Ranger Pro Sensor Interface
Datasheet

Catalog
Order
NumberPart Number

70M323XX-XX

147M7136-01-11

70M423XX-XX

147M7136-02-11

70M503XX-XXXX-XX

160M0016-01

165M4206 Rev. B

Figure(1)

25/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Drawings and Figures
Dimensions are given in mm [inches] unless noted otherwise.

Figure 1: 70M323 and 70M423 Ranger Pro Sensor Interface Modules

26/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Figure 2: Ranger Pro 70M503 Accelerometer Top Exit

27/28

Ranger Pro Sensor Interface
Datasheet

165M4206 Rev. B

Copyright 2024 Baker Hughes Company. All rights reserved.

Bently Nevada, M2, Ranger, System 1, vbOnline and Orbit Logo are registered trademarks of Bently Nevada, a Baker
Hughes business, in the United States and other countries. The Baker Hughes logo is a trademark of Baker Hughes
Company. All other product and company names are trademarks of their respective holders. Use of the trademarks
does not imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

28/28