EMS 20

User Manual (English)

EMS 20

Symbols Glossary
ISO 7000; Graphical symbols for use on equipment—Index and synopsis
Storage or operating temperature
range. Reg. # 0632
Storage humidity range Reg. # 2620
Keep away from rain, keep dry. Reg.
# 0626
Name and address of manufacturer.
Reg. # 3082
Caution, consult accompanying documents. Reg. # 0434A
Catalog Number. Reg. # 2493
Serial Number. Reg. # 2498
This way up. Reg. # 0623
Fragile, handle with care. Reg. #0621
ISO 7010: Graphical symbols—Safety colors andsafety signs—Registered safety signs
Frostbite may occur on contact withcold liquid or gaseous oxygen, or
frosted parts. Warning low temperature. To warn of low temperature or
freezing conditions. Reg. # W010
The instruction manual must be read.
Reg. # M002
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003
Do not smoke near unit or while
operating unit. Reg. # P002
Warning. Reg. # W001
ADR: European Agreement concerning the International Carriage of Dangerous Goods by Road
Non-toxic gas.
Hazard Oxidizing substances: fire
intensifying risk.
Refrigerated Liquid, USP; Produced
by Air Liquefaction

2 - ENG PN 13402391 F | User Manual

Internal Symbols
Keep unit well ventilated at all times
Keep away from flammable materials, oil and grease.
Wipe connector with clean dry cloth
before filling.
IEC 60417: Graphical Symbols for use on
equipment
Do not cover unit. These units normally vent oxygen. No. 5641
21 CFR 801.15: Code of Federal Regulations Title
21
Federal law restricts this device tosale by or on the order of a physician.
Council Directive 2012/19/EU: waste electrical andelectronic equipment (WEEE)
WEEE
This product may be covered by one or more patents, US and international. Please visit our website
below for the listing of applicable patents. Pat.:www.caireinc.com/corporate/patents/ .

EMS 20

Specifications• Mode of Operation: Continuous Flow
• Type of Protection Against Electrical Shock: Internally Powered Equipment• Equipment not suitable for use in the presence of flammable mixtures
Product Specifications

EMS 20LOX Capacity

21.0 L 50.7 lb (23 kg)

Gaseous Equivalent Capacity
Weight, Empty

17,337 L
39 lb (17,96 kg)

Weight, Filled

89.7 lb (40,69 kg)

Height24.5 in. (622 mm)

Diameter
Typical use time at 2 LPM
Operating Pressure

14 in. (356 mm)
6 days 12 hrs
50 psi (137 kPa)

Normal Evaporation Rate

1.8 lb/ day (0,82 kg/day)

PN 13402391 F | User Manual 3 - ENG

EMS 20

Warning Information
Important: Read this manual thoroughly before
operating the EMS 20.
RX Only.
WARNING: THIS DEVICE IS NOT INTENDED FOR LIFE SUSTAINING USE.
WARNING: PATIENT OR OTHERS MAY BE ENTANGLED IN CANNULA OR OTHER TUBING CAUSINGASPHYXIATION.
WARNING: IF YOU FEEL THE EQUIPMENT IS NOTOPERATING PROPERLY, CALL YOUR HEALTH
CARE PROVIDER. DO NOT ATTEMPT TO REPAIR
OR ADJUST THE UNIT YOURSELF.
WARNING: DO NOT MODIFY THIS EQUIPMENTWITHOUT AUTHORIZATION FROM THE MANUFACTURER.
WARNING: IF CONTINUITY OF OXYGEN SUPPLY IS
REQUIRED, ENSURE THAT AN ADEQUATE SUPPLYOF OXYGEN AND/OR A SECONDARY OXYGENSUPPLY IS AVAILABLE AT ALL TIMES DURING
THERAPY.
WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES WITHIN 10 FEET (3 METERS) OF
THE DEVICE, OR CLOSER THAN 8 INCHES (20 CM)
FROM ANY SOURCE OF IGNITION.
WARNING: KEEP YOUR UNIT IN A WELL-VENTILATED AREA.
WARNING: DO NOT STORE LIQUIDOXYGEN EQUIPMENT IN A CLOSET, CAR
TRUNK, OR OTHER CONFINED AREA. DO
NOT PLACE BLANKETS, DRAPERIES, OR
OTHER FABRICS OVER EQUIPMENT.
WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH IS
KNOWN TO THE STATE OF CALIFORNIA TO CAUSECANCER. FOR MORE INFORMATION, GO TO WWW.
P65WARNINGS.CA.GOV.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE, THE
USER SHOULD IMMEDIATELY REPORT THE
INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINED AS
AN INJURY, DEATH, OR POTENTIAL TO CAUSE
INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF THE INCIDENT. THE USER CAN ALSOREPORT THE INCIDENT TO THE COMPETENT AUTHORITY IN THE COUNTRY WHERE THE INCIDENTOCCURRED.

4 - ENG PN 13402391 F | User Manual

Caution: Use the EMS 20 only as directedby your doctor.
Caution: The unit contains liquid oxygen which isextremely cold, almost 300°F (-184°C). Exposure tosuch a low temperature can cause severe frostbite.
Caution: Liquid and gaseous oxygen, though nonflammable, cause other materials to burn faster thannormal. This hazard, along with the low temperatureof liquid oxygen, warrants certain safety precautions.
Caution: Keep flammable materials away from thisequipment. Aerosol sprays, oils and grease, including facial creams and petroleum jelly, ignite easily
and may burn rapidly in the presence of oxygen.
Caution: Smoking while wearing an oxygen cannula
can cause facial burns and possibly result in death.
Removing the cannula and placing it on clothing,bedding, sofas, or other cushion material will causea flash fire when exposed to a cigarette, heat source,
spark or flame.
If you smoke please: (1) turn off the portable, (2) takeoff the cannula, and (3) leave the room where thedevice is located.
Caution: In the event of an accidental tip-over, immediately but cautiously return the unit into an uprightposition if possible. If any liquid oxygen is escaping,leave the area immediately and call your healthcare
provider. Do not attempt to move the unit or stop theliquid oxygen from escaping.
Note: Do not touch frosted parts of any unit.
Note: Do not store or operate the portable coupled
to the EMS 20.
Note: Do not allow untrained personnel to handle
or operate this device.
Note: Use of this device is prohibited on commercial passenger and cargo air flights by the Federal
Aviation Administration.

Intended Use

The CAIRE EMS 20 liquid oxygen unit is intended
for the administration of supplemental oxygen. The
device is not intended for life support nor does it
provide any patient monitoring capabilities.

Introduction

Controls

The CAIRE EMS 20 unit is a liquid oxygen storage
and delivery system intended to be connected to an
ambulance’s current oxygen supply system in placeof high pressure cylinders.

1. Liquid Level Gauge

This operating instruction (OI) contains the instructions for using the EMS system, as well as information regarding available accessories. For detailed
instructions regarding accessories, please see theirrespective documentation.
This device is not meant to supply oxygen directly
to a patient. A regulation device must be in place
between the unit and the patient.

EMS 202. Liquid Level Switch
3. Output Valve
4. Fill Connection
5. Vent Valve

5

4

3

1
2

CAIRE EMS 20 shown.

PN 13402391 F | User Manual 5 - ENG

EMS 20
Operating Instructions

Basic Operations

1. Use the following chart as a guideline to determine the length of time the EMS 20 will operate:
Model
L-20
1. Remove from shipping carton and remove plugs
Off
Nominalfrom quick disconnect valve (QDV) and oxygen
0.25
34-17
gas outlet.
0.5
24-16
WARNING: THESE CONNECTION MUST BE
0.75
16-11
DRY, BECAUSE MOISTURE CAN CAUSE
1
12-8
THE EQUIPMENT TO FREEZE TOGETHERAND MAY CAUSE LEAKAGE IN THE FILL
1.5
8-5
CONNECTOR.
2
6-4
2.5
4-22
2. Position and secure EMS unit in vehicle compartment. Optional Caire mounting bracket.
3
3-2
4
2-11
3. Attach vent line from vent outlet to exterior of
5
2-1
vehicle compartment (see Accessories section for
more info).
6
1-12
8
1-5
4. Attach vehicle’s oxygen plumbing.
10
1-0
5. Attach condensate bottle bracket to EMS unit.
120-19
6. Slide condensate bottle into bottle bracket or run
15
0-19
condensate drain tube to exterior of vehicle.

Setup Instructions

7. Fill equipment with liquid oxygen and test vehicle’s oxygen delivery systems.
Note: Fill source must contain liquid oxygen saturated
between 75 psi (5.2 bar) and 100 psi (6.9 bar). Optimal
saturation pressure is 85 psi (5.9bar)

Note: Times are in days and hours (format 00-00).

8. Record location of equipment inside vehicle for
future reference.

Note: Your individual results will vary.

Fill Instructions
1. Weigh unit as required by local and federal standards.
2. Wipe reservoir fill connector and fill source withlint-free rag.
3. Open liquid use valve on fill source and purgetransfer line by pressing QDV poppet (at end oftransfer line) against non-painted surface for a
few seconds.
4. Connect pressure gauge to oxygen outlet andopen outlet valve.
5. Open vent valve on EMS unit.
6. Connect transfer line from fill source to EMS tofill. During fill, throttle the vent valve to keeppressure at oxygen outlet at approximately 50psi(3.5 bar).
7. When liquid spurts from vent outlet, perform thefollowing steps in order: close EMS vent valve,turn off outlet valve, disconnect pressure gauge,close liquid fill source, and purge transfer line.
6 - ENG PN 13402391 F | User Manual

Note: The “Nominal” times are for ideal conditions, i.e.
maximum fill, exact flow rates, good loss rate, EMS 20
not being moved, etc. These times are the maximum
expected.

2. Use the following chart as a guideline to therecommended tubing length.
FLOW
MAXIMUM (RECOMMENDED)SETTING
TUBING LENGTH*(LPM)
20 psig
50 psig
1-6
100 Ft. (30.5 m)
100 Ft. (30.5 m)
8100 Ft. (30.5 m)
75 Ft. (22.9 m)
1050 Ft. (15.2 m)
50 Ft. (15.2 m)
12
25 Ft. (7.6 m)
50 Ft. (15.2 m)
15
25 Ft. (7.6 m)
25 Ft. (7.6 m)*Length is oxygen tubing only. Does not include a 7 Ft. cannula.

EMS 20
3. Verify functionality of the gauge

To Defrost the Unit

• Depress button to display level. If level isdisplayed and Low Battery Indicator is not illuminated, battery level is acceptable.

1. Close the EMS 20 output valve and allow theunit to warm to room temperature, as indicatedby the melting of all ice from the unit.

4. To verify the level of liquid oxygen in the unit
with the liquid level gauge:

2. Check the condensation collection bottle frequently during defrosting and empty as required.

• Depress the push button on top of the unit for two
seconds minimum. Read the LED to indicate
contents level.
Caution: The EMS 20 is empty if only the
first red LED is lit.

Battery Care and Maintenance• Depress button to display level. If level isdisplayed and Low Battery Indicator is not illuminated, battery level is acceptable.

• If the Low Battery Indicator lights up when the
button is depressed, inform your health careprovider the next time your EMS 20 is filled.

• If the Low Battery Indicator lights up when the contents button
is depressed, call your Service Provider to replace the battery.

Gen 4 Meter

5. Under certain environmental conditions andwith continuous use, the EMS 20 may develop
an excessive amount of ice on the warming andbreathing coils within the shroud. You should
defrost the unit between liquid oxygen fills toprevent this ice build-up.

PN 13402391 F | User Manual 7 - ENG

EMS 20

Maintenance

Clean the fill connectors on both the stationary
and portable units with a clean, dry, lint-free cloth
between each fill to prevent freezing and possible
equipment failure.
There are no user-serviceable parts in the EMS 20s.
Your service provider is responsible for any maintenance that my be required per the technical manualof this device. Call your service provider for any
maintenance requirements.
The expected service life is a minimum of five
years.

8 - ENG PN 13402391 F | User ManualEMS 20

Troubleshooting
Issue
Inadequate Flow

Solution• Verify liquid oxygen is in unit
• Verify if cannula is kinked or pinched• Verify if cannula is properly connected to unit
NOTE: If issues persists, contact your service provider.

The liquid level meter does • The battery may need to be replaced or the meter re-calibrated. Contact service
not work or is not accurate.
provider for assistance.
The low battery LED
illuminated on liquid level
meter.
Frosting on coils of EMS
20.
There is frosting on tank orside of EMS 20.
There is a hissing soundemanating from EMS 20.

Liquid Oxygen evacuatingfrom blue QDV.

Condensation or water
pooling up on floor.
Portable takes a long time
to fill.
Portable not filling.

• Contact service provider for assistance.
• Frosting on coils is normal operation when breathing off of EMS 20.
• Frosting on outside of tank is abnormal; contact service provider for assistance.
• During normal operation the unit's primary relief valve will open from time to time torelieve excess pressure, especially soon after filling.
• If hissing is persistent or abnormal, this could indicate excess pressure being vented
off or a leak in the system. Contact service provider for assistance.
• The QDV may have frozen open. Open windows if possible and evacuate area immediately. Contact service provider.
• To prevent frozen QDV, be sure to wipe QDV with dry lint free cloth before and after
filling your portable.
• As the frost melts on the coils, water may accumulate on the floor if the condensate
bottle is not used or is full. Verify that condensate bottle is installed properly andemptied as needed.
• It could take several minutes to fill portable device if the portable device is warm or
hasn't been used recently.
• Consult user manual for your portable liquid oxygen device.
• Ensure reservoir tank has sufficient liquid to fill your portable device. Ensure portabledevice is pushed onto QDV correctly and the portable vent valve lever is held in the
open position.
• Consult user manual for your portable liquid oxygen device.

PN 13402391 F | User Manual 9 - ENG

EMS 20

Cleaning Standard

Accessories

WARNING: CLEAN ONLY AFTER THE UNIT
IS EMPTY.
• Clean using a solution of mild dish washingdetergent and water.
• Apply cleaning solution directly to a lint-free
cloth. Approved cleaners include HydroPure
and HydroKlean. Do not spray cleaners
directly on the EMS 20.
• Wipe the outside surface with the lint-free
cloth until the outside surface is clean.
Caution: Do not use high temperature andhigh pressure washing equipment to clean
these units.
• Do not get cleaner on any internal components or valves.
• Allow the unit to dry thoroughly before using.
Note: Note to health care provider – for reprocessing
procedures, see applicable service manual.

DisposalAlways return EMS 20, including all components,
to your homecare provider for proper disposal. You
can also contact your local city or town offices for
instructions on proper disposal of the battery.

WEEE

This symbol is to remind the equipment owners toreturn it to a recycling facility at the end of its life,
per Waste Electrical and Electronic Equipment (WEEE) Directive.

Transport and Storage
The device should be stored in the upright position,and be well ventilated. Do not allow the device tolie on its side. Humidity up to 95% noncondensing.
Temperatures range from -40°F to 158°F (-40°C to70°C).
Operating temperature ranges from 14°F to 104°F(10°C to 40°C). Relative humidity range from 30%
to 75% noncondensing.
Note: The atmospheric pressure range is 700 hPa to1060 hPa (elevation of 10,000 Ft. to -1,000 Ft.).

10 - ENG PN 13402391 F | User Manual• Vent Kit (PN 13399524)
Mounting Bracket (PN 13187265)

Safety

EMS 20WARNING: PORTABLE RF COMMUNICATIONS EQUIPMENT (INCLUDING PERIPHERALS SUCH AS
ANTENNA CABLES AND EXTERNAL ANTENNAS) SHOULD BE USED NO CLOSER THAN 30 CM (12INCHES) TO ANY PART OF THE EMS 20, INCLUDING CABLES SPECIFIED BY THE MANUFACTURER.
OTHERWISE, DEGRADATION OF THE PERFORMANCE OF THIS EQUIPMENT COULD RESULT.
WARNING: USE OF ACCESSORIES, TRANSDUCERS AND CABLES OTHER THAN THOSE SPECIFIED OR
PROVIDED BY THE MANUFACTURER OF THIS EQUIPMENT COULD RESULT IN INCREASED ELECTROMAGNETIC EMISSIONS OR DECREASED ELECTROMAGNETIC IMMUNITY OF THIS EQUIPMENT AND RESULT IN
IMPROPER OPERATION.
WARNING: USE OF THIS EQUIPMENT ADJACENT TO OR STACKED WITH OTHER EQUIPMENT SHOULD
BE AVOIDED BECAUSE IT COULD RESULT IN IMPROPER OPERATION. IF SUCH USE IS NECESSARY, THISEQUIPMENT AND THE OTHER EQUIPMENT SHOULD BE OBSERVED TO VERIFY THAT THEY ARE OPERATINGNORMALLY.
Caution: Medical Electrical Equipment needs special precautions regarding Electromagnetic compatibility (EMC) and needs to be installed and put into service according to the EMC information provided
in this manual.
Caution: Portable and mobile radio frequency (RF) communications equipment can affect Medical
Electrical Equipment.
Caution: The EMS 20 should not be used adjacent to or stacked with other equipment and that if
adjacent or stacked use is necessary, the EMS 20 should be observed to verify normal operation in
the configuration in which it will be used.
Table 1

Guidance and Manufacturer’s declaration—electromagnetic emissions
The EMS 20 is intended for use in the electromagnetic environment specified below. The customer or theuser of the EMS 20 should assure that it is used in such an environment.
Emissions test
Compliance Electromagnetic environment—guidance
RF emissions
Group 1The EMS 20 uses RF energy only for internal function.
CISPR 11		
Therefore, its RF emissions are very low and are not likely to			cause any interference in nearby electronic equipment.
RF emissions CISPR 11 Class B
Harmonic emissions 		
The EMS 20 is suitable for use in all establishments, including
IEC 61000-3-2
Not applicable domestic establishments and those directly connected to theVoltage fluctuations/		
public low-voltage power supply network that supplies
flicker emissions
Not applicable buildings used for domestic purposes.
IEC 61000-3-3

PN 13402391 F | User Manual 11 - ENG

EMS 20
Table 2*

Recommended separation distances between portable and mobile
RF communications equipment and the EMS 20
The EMS 20 is intended for use in an electromagnetic environment in which radiated RF disturbances
are controlled. The customer or the user of the EMS 20 can help prevent electromagnetic interference by
maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the EMS 20 as recommended below, according to the maximum output power of the communications equipment.
Rated maximum output
Separation distance according to frequency of transmitter
power of transmittermW
150 kHz to 80 MHz
80 MHz and 800 MHz
800 MHz to 2,5 GHz		
d=1.2√P
d=1.2 √P
d=2.3 √P
0,01
0.12 m0.12 m
0.23 m0,10.38 m0.38 m0.73 m
1
1.2 m
1.2 m
2.3 m
10
3.8 m
3.8 m
7.3 m
100
12 m
12 m
23 m
For transmitters rated at a maximum output power not listed above, the recommended separation distance(d) in meters (m) can be estimated using the equation applicable to the frequency of the transmitter,
where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter
manufacturer.
NOTE 1 at 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected byabsorption and reflection from structures, objects and people.
* This table is included as a standard requirement for equipment which has been tested to specific testlevels and over specific frequency ranges and been found compliant with regulations.
Table 3

Guidance and manufacturers declaration—electromagnetic immunity
The EMS 20 is intended for use in the electromagnetic environment specified below. The customer orthe user of the EMS 20 should assure that it is used in such an environment.
Immunity test
Electrostatic
discharge (ESD)
IEC 61000-4-2
Electrical fasttransient/burst
IEC 610004-4
		Surge
IEC 61000-4-5

IEC 60601 test level
Compliance level
Electromagnetic environment—guidance+- 8kV Contact+- 8kV Contact
Floors should be wood, concrete or ceramic+- 2 kV, +-4 kV,+- 2 kV, +-4 kV,
tile. If floors are synthetic, the relative+- 8 kV, +- 15 kV air+- 8 kV, +- 15 kV air humidity should be at least 30%.**±2 kV for powerNot applicableNot applicablesupply lines
DC powered device±1 kV forNot applicable		
input/output lines
No data input/output lines±1 kV line(s) to line(s)Not Applicable
Not Applicable±2 kV line(s) to earth
DC powered device

Voltage dips,<5% UT (>95% dip in UT)		
short interruptions
for 0,5 cycle		and voltage
40% UT (60% dip in UT)
Not Applicablevariations on
for 5 cycles
DC powered
power supply
70% UT (30% dip in UT)
deviceinput lines
for 25 cycles		IEC 61000-4-11 <5% UT (>95% dip in UT) for 5 sec
Power frequency(50/60 Hz)
magnetic field
IEC 61000-4-8

30 A/m30 A/m50/60 Hz50/60 Hz		

Not Applicable		Power frequency magnetic fields should bethat of a typical commercial or hospitalenvironment.

Note: UT is the a.c. mains voltage prior to application of the test level.

** This statement indicates that the required testing was performed in a controlled environment and the
EMS 20 are found to be compliant with regulations.
12 - ENG PN 13402391 F | User ManualEMS 20
Table 4
Guidance and Manufacturer’s Declaration—Immunity ME Equipment and ME Systems
Guidance and Manufacturer’s Declaration—Immunity
The EMS 20 is intended for use in the electromagnetic environment specified below. The customer oruser of the EMS 20 should ensure that it is used in such an environment.
Immunity Test

IEC 60601 Test Level

Compliance Level

Electromagnetic Environment—Guidance

Conducted RF IEC
61000-4-6

3 Vrms6 Vrms (In ISM Bands)
150 kHz to 80 MHz

Not applicable

Portable and mobile RF communications equipment should be used no closer to any part of
the EMS 20, including cables, than the recommended separation distance calculated fromthe equation applicable to the frequency of thetransmitter.

Battery powered
device, No SIP/SOP

Recommended separation distanced = 1,2 √P
Radiated RF IEC
61000-4-3

80 MHz to 2.7 GHz

10 V/m80 MHz—2,7 GHz
80 % AM at 1 kHz

d = 1,2 √P
d = 2,3 √P
where P is the maximum output power ratingof the transmitter in watts (W) according to thetransmitter manufacturer and d is the recommended separation distance in meters (m).
Field strengths from fixed RF transmitters, as
determined by an electromagnetic site surveya,
should be less than the compliance level in eachfrequency rangeb.
			Interference may occur in the vicinity of equipment marked with the following symbol:

Test frequency(MHz)

Banda)(MHz)

Servicea)

Modulationb)

385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

Pulse modulationb)
18 Hz
FMC)
±5 kHz deviation
1 kHz sine

Maximum
power (W)Distance(m)

Immunity Test
Level (V/m)

1.8

0.3

27

2

0.3

28

710
745
780
810
870
930
1720
1845
1970

704–787

LTE Band 13, 17

Pulse modulationb)
217 Hz

0.2

0.3

9

800–960

GSM 800/900, TETRA
800, IDEN 820, CDMA
850, LTE Band 5

Pulse modulationb)
18 Hz

2

0.3

28

1700–
1900

GSM 1800; CDMA 1900;
GSM 1900; DECT; LTEBand 1, 3, 4, 25; UMTS

Pulse modulationb)
217 Hz

2

0.3

28

2450

2400–
2570

Pulse modulationb)
217 Hz

2

0.3

28

5240
5500
5785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, LTE
Band 7

5100–
5800

WLAN 802.11 a/n

Pulse modulationb)
217 Hz

0.2

0.3

9

NOTE: If necessary to achieve the IMMUNITY TEST LEVEL, the distance between the transmitting antenna and the
EMS 20 may be reduced to 1m. The 1m test distance is permitted by IEC 61000-4-3.
For some services, only the uplink frequencies are included.
The carrier shall be modulated using a 50% duty cycle square wave signal.
As an alternative to FM modulation, 50% pulse modulation at 18 Hz may be used because while it does not represent actual modulation, it
would be worst case.
abc

PN 13402391 F | User Manual 13 - ENG

EMS 20Notes

14 - ENG PN 13402391 F | User ManualNotesEMS 20

PN 13402391 F | User Manual 15 - ENG

www.ca ire inc .c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
Email: info@mps-gmbh.eu
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please visit our
website below for a full listing of trademarks. Trademarks: www.caireinc.com/
corporate/trademarks/.
Copyright © 2022 CAIRE Inc. CAIRE Inc. reserves the right to discontinue
its products, or change the prices, materials, equipment, quality, descriptions,specifications and/or, processes to its products at any time without prior notice andwith no further obligation or consequence. All rights not expressly stated herein arereserved by us, as applicable.

|13402391-F[~
3 May 2022

13402391 F