USER’S MANUAL

DaqBook/2000 Series and
DaqOEM/2000 Series
16-Bit, 200 kHz Multifunction Data Acquisition Systems
DaqBook/2001
DaqBook/2005
DaqBook/2020
DaqOEM/2001
DaqOEM/2005

Requires one of the following
Operating Systems:

Windows XP
Windows Vista

DaqBook/2000 Series &DaqOEM/2000 Series

Measurement Computing

1121-0901 rev 3.0

10 Commerce Way
Norton, MA 02766*372335A-01*(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

372335A-01

www.mccdaq.com

ii

Warranty Information
Contact Measurement Computing by phone, fax, or e-mail in regard to warranty-related issues:
Phone: (508) 946-5100, fax: (508) 946-9500, e-mail: info@mccdaq.com

Limitation of Liability
Measurement Computing cannot be held liable for any damages resulting from the use or misuse of this product.

Copyright, Trademark, and Licensing Notice
All Measurement Computing documentation, software, and hardware are copyright with all rights reserved. No part ofthis product may be copied, reproduced or transmitted by any mechanical, photographic, electronic, or other method
without Measurement Computing’s prior written consent. IOtech product names are trademarked; other product names, asapplicable, are trademarks of their respective holders. All supplied IOtech software (including miscellaneous support
files, drivers, and sample programs) may only be used on one installation. You may make archival backup copies.

CE Notice
Many Measurement Computing products carry the CE marker indicating they comply with the safety and emissions
standards of the European Community. When applicable these products have a Declaration of Conformity stating which
specifications and operating conditions apply. You can view the Declarations of Conformity at
www.mccdaq.com/legal.aspx (CE Information page).

Warnings, Cautions, Notes, and Tips
Refer all service to qualified personnel. This caution symbol warns of possible personal injury or equipment damage
under noted conditions. Follow all safety standards of professional practice and the recommendations in this manual.
Using this equipment in ways other than described in this manual can present serious safety hazards or cause equipment
damage.
This warning symbol is used in this manual or on the equipment to warn of possible injury or death from electricalshock under noted conditions.

This ESD caution symbol urges proper handling of equipment or components sensitive to damage from electrostatic
discharge. Proper handling guidelines include the use of grounded anti-static mats and wrist straps, ESD-protective bags
and cartons, and related procedures.

This symbol indicates the message is important, but is not of a Warning or Caution category. These notes can be of great
benefit to the user, and should be read.
In this manual, the book symbol always precedes the words “Reference Note.” This type of note identifies the locationof additional information that may prove helpful. References may be made to other chapters or other documentation.
Tips provide advice that may save time during a procedure, or help to clarify an issue. Tips may include additionalreference.

Specifications and Calibration
Specifications are subject to change without notice. Significant changes will be addressed in an addendum or revision to
the manual. As applicable, the hardware is calibrated to published specifications. Periodic hardware calibration is notcovered under the warranty and must be performed by qualified personnel as specified in this manual. Improper
calibration procedures may void the warranty.

Your order was carefully inspected prior to shipment. When you receive your system, carefully
unpack all items from the shipping carton and check for physical signs of damage that may haveoccurred during shipment. Promptly report any damage to the shipping agent and your sales
representative. Retain all shipping materials in case the unit needs returned to the factory.

CAUTION
Using this equipment in ways other than described in this manual can cause
personal injury or equipment damage. Before setting up and using yourequipment, you should read all documentation that covers your system.
Pay special attention to Warnings and Cautions.
Note:

During software installation, Adobe® PDF versions of user manuals will automaticallyinstall onto your hard drive as a part of product support. The default location is in the
Programs group, which can be accessed from the Windows Desktop. Initial navigation
is as follows:
Start [Desktop “Start” pull-down menu]⇒ Programs⇒ IOtech DaqX Software
You can also access the PDF documents directly from the data acquisition CD by using
the <View PDFs> button located on the opening screen.
Refer to the PDF documentation for details regarding both hardware and software.
A copy of the Adobe Acrobat Reader® is included on your CD. The Reader provides
a means of reading and printing the PDF documents. Note that hardcopy versions of
the manuals can be ordered from the factory.

DaqBook2000_DaqOEM2000.pdf
PDF

1121-0901

Contains hardware-related and software-related material for DaqBook/2000,
DaqBook/2001, DaqBook/2005, DaqBook/2020, DaqOEM/2001, and DaqOEM/2005.
This pdf file plus the following constitute a complete set of documentation for thesedevices.

DaqView.pdf

Discusses how to install and use this “out-of-the-box” data acquisition program.
PDF

457-0909

PostAcquisition Analysis.pdf
PDF

1086-0926
1086-0922

iv

This pdf consists of two documents. The first discusses eZ-PostView, a post dataacquisition analysis program. The application is included free as a part of productsupport. The second includes information regarding eZ-FrequencyView andeZ-TimeView. These two applications have more features than does eZ-PostView
and are available for purchase. They can; however, be used freely during a 30-day
trial period.

898695

DaqBook/2000 Series and DaqOEM/2000 Series

DBK Options.pdf
PDF

The DBK Option Cards and Modules User’s Manual discusses each DBK product.
In addition, the document includes chapters on signal management, system
connections, pinouts, DBK set up in software, and troubleshooting.

457-0905

ProgrammersManual.pdf

The Programmer’s Manual pertains to developing custom programs using
Applications Program Interface (API) commands.
PDF

1008-0901

Programmers should check the readme.file on the install CD-ROM for the location ofprogram examples included on the CD.

Manual Layout
Chapter 1 – Daq Systems
Chapter 2 – DaqBook/2000 Series and DaqOEM/2000 Series Device Overviews
Chapter 3 – Block Diagrams
Chapter 4 – Connectors, Indicators, Cables, & Pinouts
Chapter 5 – A Brief Look at Software
Chapter 6 – Configuring the System for Operation
Chapter 7 – CE Compliance and Noise Considerations
Chapter 8 – Taking Measurements
Chapter 9 – Calibration
Chapter 10 – Troubleshooting and Customer Support
Chapter 11 – Specifications
Glossary
Reference Note:
Reference to additional documents of importance begins on page iv.

DaqBook/2000 Series and DaqOEM/2000 Series

898695v

This page is intentionally blank.

vi

898695

DaqBook/2000 Series and DaqOEM/2000 Series

Table of Contents
Quick Start
1 – Daq Systems
2 – DaqBook/2000 Series and DaqOEM/2000 Series, Device Overviews
I/O Comparison Matrix …… 2-1
Overview …… 2-2
Product Features …… 2-2
Synchronous Operations Matrix …… 2-6
DaqBook/2001 and DaqOEM/2001 …… 2-7
DaqBook/2005 and DaqOEM/2005 …… 2-8

3 – Block Diagrams
DaqBook/2001, DaqOEM/2001, DaqBook/2005, & DaqOEM/2005 …… 3-1
DaqBook/2020 …… 3-2
Detail “TC” (Thermocouple Input Circuitry) for DaqBook/2020 …… 3-3
Detail “V” (Differential Voltage Input Circuitry) for DaqBook/2020 …… 3-3

4 – Connectors, Indicators, Cables, & Pinouts
DaqBook/2020 …… 4-1
DaqBook/2001 and DaqBook/2005 ….. 4-3
DaqOEM/2001 and DaqOEM/2005 …… 4-4
Cables …… 4-5
Pinouts …… 4-7

5 – A Brief Look at Software
6 – Configuring the System for Operation
7 – CE Compliance and Noise Considerations
8 – Taking Measurements
Introduction …… 8-1
Scanning …… 8-1
Triggering …… 8-2
Synchronous I/O Operations …… 8-3
Asynchronous I/O Operations …… 8-7
Using DBK Cards and Modules for Signal Conditioning …… 8-8

9 – Calibration
10 – Troubleshooting and Customer Support
Electrostatic Discharge (ESD), Handling Notice…… 10-1
Product Care …… 10-1
ReadMe Files and the Install CD-ROM …… 10-2
Driver Support…… 10-2
DaqCOM Issues …… 10-2
Ethernet Problems …… 10-2
Frequently Asked Questions …… 10-6
Customer Support …… 10-9

11 - Specifications
Glossary

DaqBook/2000 Series and DaqOEM/2000 Series

999895vii

This page is intentionally blank.

viii

999895

DaqBook/2000 Series and DaqOEM/2000 Series

DaqBook & DaqOEM

Quick Start

2000 Series

Ethernet-Based Portable Data Acquisition
Before you get started
DaqBook/2020

Verify that you have the following items.

DaqOEM/2000

DaqBook/2001

•
•
•
•
•
•
•

DaqBook, or DaqOEM [2000 Series]
TR-40U Power Supply
Ethernet Patch Cable
Ethernet jack [on PC or on a hub connected to the Ethernet]

Data Acquisition CD
Monitor: SVGA, 1024 x 768 screen resolution

Windows 2000 and Windows XP users:
PC with Intel™ Pentium, 1 GHz or equivalent;
512 MB memory; 10 GB disk space• Windows Vista users:
PC must be Windows Vista Premium Ready

Step 1 - Install Software
1.

Close all running applications on the host PC.

2.

Insert the Data Acquisition CD into your CD-ROM drive and wait for the CD to auto-run. An Opening Screen
will appear. As an alternative, you can download software from: ZZZPFFGDTFRPVRIWZDUH

3.

Click the <ENTER SETUP> button.
Note: If you are downloading software from our website, follow instructions provided there.

4.

From the hardware selection screen [which follows a licensing agreement], select the device from thedrop-down list and follow the on-screen instructions.

Step 2 - Connect the 2000 Series Device to the Ethernet
Though the following figure represents connecting a DaqBook/2020 to a PC and to power, note that the Ethernet ports
and Power In connectors are in the same relative locations for all devices in this 2000 Series.

In the figure, the DaqBook/2020 is connected directly to an Ethernet port on a host computer. Please consult youruser’s manual (located in PDF format on the CD) should you need information regarding network types other than thissimple direct connection.
1.

Connect the Ethernet cable to the Ethernet jack on the 2000 Series device rear panel.

2.

Connect the other end of the Ethernet cable to the Ethernet jack on the host computer or
network hub.

Step 3 - Connect the 2000 Series Device to Power
1.

With the 2000 Series power switch “OFF,” connect the power supply cable from the TR-40U to the unit’s
Power-In DIN5 connector (located on the rear panel).

2.

Connect the TR-40U plug to a standard AC outlet; then turn the 2000 Series device Power switch to “ON.”1121-0940, rev 1.0

324631A-01

Printed in Hungary

Step 4 - Configure Computer Network Settings
Note: We recommend that you discuss this procedure with your Network Administrator before proceeding.

1.
2.
3.
4.
5.
6.
7.

8.
9.

Open the Control Panel by navigating from the Windows Desktop as follows:
Start Menu ⇒ Settings ⇒ Control Panel.
Double-click the “Network Connections” icon.
Double-click the icon for the network that the 2000 Series device is connected to.
In the “Local Area Connection Status” box, click on the <Properties> button. The “Local Area Connection
Properties” box will appear.
Double-click the “Internet Protocol (TCP/IP)” component. The “Internet Protocol (TCP/IP) Properties” boxwill appear.
Select the “Use the following IP Address” radio
button. (See figure at left).
Set the IP address field to 10.0.0.x
where x is some number from 1 to 254.
Make sure that each computer and each device on
the dedicated network has a unique IP address.
Set the Subnet mask to 255.0.0.0. Note that theremaining fields can be left as is. Click <OK>.
Internet Protocol (TCP/IP) Properties
Click <OK> on follow-up screens to exit.
(Partial View)

Step 5 - Configure & Test the System
1.

Open the Daq Configuration Applet.
a.
b.

Navigate from the Windows’ Desktop: Start Menu ⇒ Settings ⇒ Control Panel
From the Control Panel, double-click the Daq Configuration icon.

2. Add the 2000 Series Device to the list of installed devices.
a.
b.
c.

Select the Computer image in the Device Inventory configuration tree.
Click the <Add Device> button. The “Select Device Type” box will appear.
Select the 2000 Series device from the list of devices.
Note: For DaqOEM/2001 and DaqOEM/2005, select DaqBook/2001 and DaqBook/2005, respectively.

d.
e.
f.
g.

Click the <OK> button. The “Properties” box will appear for the selected device.
Enter the Serial Number of the device. The number is on the MAC label (rear panel, upper left). The
serial number is located just below a barcode.
Select the “Auto IP Setting” radio button. The IP Address of the device will automatically becalculated and displayed in the IP Address field
Click the <OK> button.

3. Test the system connections.
a.
b.
c.
d.
e.

Make sure the device has been properly installed and is powered-on.
Make sure all cables are properly and securely connected.
Click the “Test Hardware” tab.
Click the <TCP/IP Test> button. This tests the Transmission Control Protocol / Internet Protocol.
Upon completion of the TCP/IP test, click the <Resource Test> button.
When testing, if the unit does not respond within 30 seconds perform the following steps:
1) reboot the system, 2) upon power-up, re-open the Daq Configuration applet, 3)select anotherconfiguration setting, 4) reinitiate the test.

Step 6 - Connect Data Acquisition Signal Lines
Prior to making signal connections review the Specifications chapter of your user’s manual to ensure that yourintended signal inputs do not exceed the specified limits. The manual is included in PDF format on your CD.

Measurement Computing
10 Commerce Way
Norton, MA 02766(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

www.mccdaq.com*324631A-01*
324631A-01

Printed in HungaryDaq Systems

1

Daq Systems, the Modular Concept …… 1-1
DaqBooks, DaqBoards, and Daq PC-Cards ….. 1-2
Using DBK Cards and Modules for Signal Conditioning …… 1-4
Daq Software …… 1-5

This chapter provides a brief overview of Daq devices.
Users interested in jumping straight to specific DaqBook/2000 Series or DaqOEM/2000 Series
information can skip directly to chapter 2.

Daq Systems, the Modular Concept
Daq equipment and software form a modular, interrelated family of products that provide great flexibility
in data acquisition system design. This flexibility allows for the development of custom systems that are
unique to the user, and which can be optimized for his or her specific application needs. With the Daq
product line, system expansion or redesign can typically be accomplished with relative ease.
•

Primary Acquisition Device. This is the main data acquisition device, e.g., a DaqBook, DaqBoard,or Daq PC-Card. These devices provide a vital data conversion and communications link between
the data source of transducers and signal conditioners and the data processor of the host computer.

•

DBK Option Cards and Modules. Over 35 DBK cards and modules (the number is constantly
growing) provide various types of signal conditioning and system expansion. Note that certain DBK
modules exist for the purpose of supplying power to other members of the acquisition system. The
DBK options are discussed in the DBK Option Cards and User’s Manual (p/n 457-0905).
Note: Only passive DBKs, such as the DBK1 BNC module, the DBK11A screw terminal card, andthe DBK40 BNC analog interface, can be used with a Daq PC-Cards.
Reference Note:
DBK options are discussed in the DBK Option Cards and Modules User’s Manual(p/n 457-0905). As a part of product support, this manual is automatically loaded onto
your hard drive during software installation. The default location is the Programs
directory, which can be accessed through the Windows Desktop.

•

Software. DaqView out-of-the-box software provides a graphical user interface with easy to readspreadsheet formats for viewing channel data, as well as a choice of analog, digital, and bar-graphmeters. Waveform analysis can be performed, when applicable. A product support option, included
on the data acquisition CD, provides a means of performing post data analysis. More information is
included in the software-specific PDF documents that are installed on your hard-drive as a part ofproduct support.
In addition to the included out-of-the-box software, Daq products can be controlled via user-written
custom programs through Applications Program Interface (API). Several languages are supported,e.g., C/C++, and VisualBASIC.
Note that third party icon-driven software, such as DASYLab and LabView, can be used with the
Daq devices.

DaqBook/2000 Series and DaqOEM/2000 Series

958795

Daq Systems

1-1

DaqView and DASYLab can only be used with one DaqBook/2000 Series, DaqOEM
Series, or DaqBoard/2000 Series device at a time.
LabView can be used with multiple /2000 Series devices.
For multiple device applications (via custom programming) refer to the Using
Multiple Devices section of the Programmer’s Manual. During software installationfrom the data acquisition CD (p/n 1022-0601), a PDF version of the Programmer’s
Manual is automatically loaded onto your hard drive as a part of product support.
The default location is the Programs directory.
Reference Note:
Programming topics are covered in the Programmer’s User Manual (p/n 1008-0901).
As a part of product support, this manual is automatically loaded onto your hard drive
during software installation. The default location is the Programs directory, which canbe accessed through the Windows Desktop.

DaqBooks, DaqBoards and Daq PC-Cards
Daq products connect to one or more DBKs on their signal input side and a computer on their output side.
Each type of Daq device connects to the computer in a different way:

1-2

Daq Systems•

A DaqBook/2000 Series product is a primary data acquisition device which resides external to a PC.
These devices connect directly to computer’s Ethernet port, or to an Ethernet network via a hub.
Note that early DaqBook models connected to a computer’s enhanced parallel port (EPP) interface orconnected to the PC by use of a PC-Card and cable.

•

DaqOEM/2000 Series devices are DaqBook/2000 Series devices, minus the chassis.

•

The DaqBoard [ISA type] board is an internal card that plugs into an ISA-bus slot within acomputer.

•

DaqBoard/2000 Series boards plug into a PCI-bus slot, within a host PC. A variation of theseboards, the DaqBoard/2000c Series boards, plug into a compact-PCI-bus slot, within a compact
type host PC.

•

The Daq PC-Card slides into the PCMCIA slot of a host computer, typically a notebook PC.

958795

DaqBook/2000 Series and DaqOEM/2000 Series

CategoryPrimary
Acquisition
DeviceDBK Option
Cards and
Modules*

Software

Daq Data Acquisition Devices
DeviceDescription
DaqBook/2000
SeriesPortable Data Acquisition

DaqOEM/2000
SeriesPortable Data Acquisition

DaqBook/100 Series
DaqBook/200 Series

Portable Data Acquisition
12-bit: DaqBook/100, /112, /120
16-bit: DaqBook/200, /216, /260

DaqBoard/2000
SeriesPlug-In Boards for PCI Bus-Slots
16-bit , 200 kHz. Six boards identified as /2000 through /2005

DaqBoard/2000c
SeriesPlug-In Boards for Compact-PCI Bus-Slots
16-bit , 200 kHz. Six boards identified as /2000c through/2005cDaqBoard (ISA
types)*

Plug-In Boards for ISA Bus-Slots
12-bit: DaqBoard/100A, /112A
16-bit: DaqBoard/200A, /216A, /2000

Daq PC-Card*

Plug-In PCMCI Card
12-bit: Daq/112B
16-bit: Daq/216B

Analog Signal
Conditioning

Cards and DBK modules used to condition Analog Signals
DBK/ 4, 7, 8, 9, 12, 13, 15, 17, 18, 19, 42, 43A, 44, 45, 50, 51,
52, 53, 54, 55, 65, 80, 81, 82, 83, 84, 85, 90, 100, 207,
207/CJC

Analog Output

Cards used to modify Analog Output Signals
DBK/ 2, 5

Digital I/O andControlCards and DBK modules used to condition Digital I/O
DBK/ 20, 21, 23, 24, 25, 208, 210

Expansion
Connections

Cards and DBK modules used to expand the acquisition
system.
DBK/ 1, 10, 11A, 35, 40, 41, 60, 200, 201, 202, 203, 204, 205,
206, 209

16-bit, 200 kHz devices with Ethernet connectivity. The
series consists of DaqBook/2001, DaqBook/2005, and
DaqBook/2020.
16-bit, 200 kHz boards with Ethernet connectivity. The
series consists of DaqOEM/2001 and DaqOEM/2005.
These boards reside external to the host PC.

Power Supply

DBKs: 30A, 32A, 33, 34

Included Software

DaqView, Post Data Acquisition Analysis Program (actual
application not specified), Visual Basic extensions, Application
Programming Interface (API)

Optional Software

DaqViewXL, DASYLab* The list of DBK cards and modules continues to grow. For information regarding new and upcoming products
consult out website, your sales representative, or contact the factory directly.

DaqBook/2000 Series and DaqOEM/2000 Series

958795

Daq Systems

1-3

Using DBK Cards and Modules for Signal Conditioning
The DBK signal-conditioning cards and modules are designed for use with DaqBooks, LogBooks, andvarious types of data acquisition boards, i.e., ISA, PCI, and compact-PCI types. The DBKs perform bestwhen used with an acquisition device that can dynamically select channel, gain, and range. DBK cards andmodules with dynamic channel and gain/range selection allow for high channel-to-channel scan rates witha variety of transducers.
Note: Only passive DBKs, such as the DBK1 BNC module, the DBK11A screw terminal card, and the
DBK40 BNC analog interface, can be used with Daq PC-Cards.
DBK output signals can be bipolar, e.g., -5 to +5 V, or unipolar, e.g., 0 to 10 V. The user can select a
range of relevant values to correspond to the lowest signal (e.g., -5 or 0 V) and the highest signal (e.g., 5 or
10 V) signal. This type of range selection guarantees the highest resolution in 12-bit or 16-bit conversion.
DBK modules share the same footprint as the DaqBook and a typical notebook PCs; allowing forconvenient stacking. The majority of these modules have their own power supply; however, several
options exist for packaging and powering the DBKs.
Reference Note:
DBK options are detailed in the DBK Option Cards and Modules User’s Manual(p/n 457-0905). As a part of product support, this manual is automatically loaded onto yourhard drive during software installation. The default location is the Programs directory, which
can be accessed through the Windows Desktop.

1-4

Daq Systems

958795

DaqBook/2000 Series and DaqOEM/2000 SeriesDaq Software
The Daq devices have software options capable of handling most applications. Three types of software areavailable:• ready-to-use graphical programs, e.g., DaqView, DaqViewXL, and post acquisition data analysis
programs such as eZ-PostView• drivers for third-party, icon-driven software such as DASYLab and LabView• various language drivers to aid custom programming using API
Ready-to-use programs are convenient for fill-in-the-blank applications that do not require programmingfor basic data acquisition and display:•

DaqView is a Windows-based program for basic set-up and data acquisition. DaqView lets you
select desired channels, gains, transducer types (including thermocouples), and a host of otherparameters with a click of a PC’s mouse. DaqView lets you stream data to disk and display data
in numerical or graphical formats. PostView is a post-acquisition waveform-display program
within DaqView.

•

DaqViewXL allows you to interface directly with Microsoft Excel to enhance data handling anddisplay. Within Excel you have a full-featured Daq control panel and all the data display
capabilities of Excel.

•

A post acquisition data analysis programs, e.g., eZ-PostView, typically allows you to view andedit post-acquisition data.

•

The Daq Configuration control panel allows for interface configuration, testing, andtroubleshooting.

Each Daq system comes with an Application Programming Interface (API) via a subroutine library (DAQ)or an ActiveX/COM based library (DaqCOM). API-language drivers include: C/C++, and Visual Basic.
The latest software is a 32-bit version API.

Reference Notes:
¾
¾

PDF
Note:

The software document modules, DaqView, DaqViewXL, and Post Acquisition Data
Analysis User’s Guide, are not included as part of the hardcopy manual, but areavailable in PDF version. See the PDF Note, below.
Programming topics are covered in the Programmer’s User Manual (1008-0901). As apart of product support, this manual is automatically loaded onto your hard drive duringsoftware installation. The default location is the Programs directory, which can beaccessed through the Windows Desktop.

During software installation, Adobe® PDF versions of user manuals will automaticallyinstall onto your hard drive as a part of product support. The default location is in the
Programs directory, which can be accessed from the Windows Desktop. Refer to the PDF
documentation for details regarding both hardware and software.
A copy of the Adobe Acrobat Reader® is included on your CD. The Reader provides
a means of reading and printing the PDF documents. Note that hardcopy versions of themanuals can be ordered from the factory.

DaqBook/2000 Series and DaqOEM/2000 Series

958795

Daq Systems

1-5

1-6

Daq Systems

958795

DaqBook/2000 Series and DaqOEM/2000 Series

DaqBook/2000 Series and DaqOEM/2000 Series

2

Device Overviews
I/O Comparison Matrix …… 2-1
Overview …… 2-2
Product Features …… 2-2
Synchronous Operation Matrix …… 2-6
DaqBook/2001 and DaqOEM/2001 …… 2-7
DaqBook/2005 and DaqOEM/2005 ….. 2-8

I/O Comparison Matrix
for DaqBook/2000 Series and DaqOEM/2000 Series Devices

DeviceDaqBook/2001
andDaqOEM/2001

DaqBook/2005
andDaqOEM/2005

Analog Input
Channels

Analog Output
Channels

Digital I/O
Channels*

Counter/
TimersAnalog Input signals
enter through P1 , go
to MUX, to PGA, toGain & Offset
Amplifier, then to
Analog-to-Digital
Converter (ADC)

Digital Signals go
through Digital-toAnalog Converters,then through “DAC
Out.”

Digital signals passthrough one 16-bit
Digital I/O Port (onP3) and three 8-bit
Digital I/O Ports(on P2).

Four 16-bit Counter
Input signals and Two
16-bit Timer Output
signals.

16 Analog
Input
via P1

4 Analog
Outputvia P316 Analog
Input via P1

No DAC
Channels

14 Analog
Input via P1

4 Analog
Outputvia BNC
Connectors

DaqBook/2020
16 Differential
Voltage Inputs
via BNC
Connectors

14 Thermocouple
Inputsvia Mini-TC
Connectors

40 Digital
I/O
via P2 & P3

40 Digital
I/O
via P2 & P3

24 Digital
I/O via P2

4 Counter In
2 Timer Out
via P34 Counter In
2 Timer Out
via P34 Counter In
2 Timer Out
via BNC
Connectors

Note: DaqBook/2020’s Digital I/O is via the three 8-bit ports on
P2. The device has no P3 connector. For this reason it has
24 Digital I/O channels available, not 40 as do the otherdevices.

Note: In addition to the BNC connectors indicated above, the DaqBook/2020 includes a BNC for External TTL Trigger input
and one for External Clock input.
DaqBook/2000 Series and DaqOEM/2000 Series

898095

Device Overviews

2-1

Overview
DaqBook/2020, /2001, /2005, DaqOEM/2001, and DaqOEM/2005 are high-speed, multi-function, dataacquisition devices for use with Ethernet PCs and Ethernet networks. Each unit connects directly to a PC’s
Ethernet port or network hub, and each features a 16-bit, 200-kHz A/D converter and digital calibration.
DaqOEM/2001 and DaqOEM/2005 devices have no chassis, but are otherwise identical to the
DaqBook/2001 and DaqBook/2005, respectively.
Note that block diagrams are presented in chapter 3 and connectors and pinouts are discussed chapter 4.

Product Features
I/O Connections
For DaqBook/2020, all input and output signals are accessed via: 37-pin DSUB connectors [P1 and P2],
BNC Connectors, and mini-TC connectors. Chapter 4 includes connector descriptions and pinouts.
For DaqBook/2001, /2005 and DaqOEM/2001, and DaqOEM/2005 all input and output signals areaccessed via the three 37-pin DSUB connectors [P1, P2, and P3]. Chapter 4 includes connector
descriptions and pinouts.

General Features•

TTL-level digital I/O lines. For DaqBook/2020 there are 24 TTL-level digital I/O lines
divided into three 8-bit ports on P2. For the other /2000 Series devices there are 40 digital
I/O lines. They are divided into three 8-bit ports on P2 and one 16-bit port on P3.

•

Four 16-bit counters. Each can accept frequency inputs up to 10 MHz. The counters can becascaded into two 32-bit counters.

•

Two 16-bit timer outputs. Each can generate square waves from 16 Hz to 1 MHz.

•

Configuration through software only. Device settings are made via software. There are no
hardware settings to be made, for example, there are no device DIP switches to be set or
jumpers to be positioned. However, note that DBK expansion options may require s small
degree of manual set up. Refer to the DBK Option Cards and Modules User’s Manual(457-0905) for details. A PDF version of the document is included on the installation CD.

•

Scan Sequencing. Each DaqBook/2020, /2001, /2005, DaqOEM/2001, and DaqOEM/2005
has a scan sequencer lets you select up to 16384 channel/range combinations. Scan rates areprogrammable on a per channel basis to 5 µs/channel, 10 µs/channel, or 1000 µs/channel.
Note: Inputs to these devices can be measured at the 5 µs/channel rate. However, some
DBK expansion options require more settling time and can be measured at theslower rate of 10 µs/channel or 1000 µs/channel. All types of measurements canbe mixed within the scan group.

•

2-2

Device Overviews

Synchronization. Each device includes two SYNC ports. Up to four units can besynchronized, with respect to each other. Synchronization is accomplished by connecting one
DaqBook device SYNC port to the SYNC port of a second DaqBook device. A unit can be
scan-synchronized and triggered from any other DaqBook that is connected to it via a SYNC
cable. Either one of a unit’s two sync ports, or both, can be used. Note that the total
combined length of the Sync cables is not to exceed 15 feet (4.57 m). Chapter 8 includes a
block diagram regarding synchronizing multiple units.

898095

DaqBook/2000 Series and DaqOEM/2000 Series

Triggering
DaqBook/2020, /2001, /2005, DaqOEM/2001, and DaqOEM/2005 each support a full complement oftrigger modes including:•

Hardware analog triggering – A user-programmed trigger level sets an analog DAC, whichis compared in hardware to the analog input level on the selected channel. Trigger latencyis < 5 µs.

•

Digital and pattern triggering – TTL-level triggering and latencies less than 5 µs. The
trigger can be programmed for logic level or edge triggering. In pattern triggering, any of thedigital input ports acts as the trigger port. You can program the digital pattern.

•

Software-based triggering – The PC detects the trigger event from analog, digital, or
counter readings. Six pre- and post-triggering modes are supported. Trigger latency is one
scan period.
Multi-unit synchronization via the SYNC ports does not support pre-trigger data.
Only post-trigger data can be collected and time correlated between the units.

Synchronous Input Operations
The DaqBook/2000 Series and DaqOEM/2000 Series devices allow for synchronous scanning andacquisition of Analog Input, Digital Input and Counter Input Data at up to 200 kHz aggregate scanning
rates. The Analog Input data can be either main unit or expansion modules from P1 compatible analog
input modules. The Digital Input data can be main unit 8-bit P2 (8255) digital inputs, 16-bit P3* digital
inputs or P2 compatible DBK digital input expansion modules.
*The 16-bit P3 digital inputs do not apply to DaqBook/2020, which has no P3 connector.
Analog Input Channels

The DaqBook/2000 Series and DaqOEM/2000 Series devices allow analog input for the device and for P1
compatible DBK analog input expansion modules.
Channel Selection and Mode Settings

The DaqBook/2000 Series and DaqOEM/2000 Series devices each accept up to 16 single ended or up to
8 differential-ended inputs and can be programmed for single-ended or differential-ended on a per channel
basis. Just one analog channel is sacrificed when a DBK expansion module is enabled. See DBK
documentation in the DBK Option Cards & Modules User’s Manual (p/n 457-0905) for further
information.
The DaqBook/2020 accepts up to 14 single ended inputs.
Channel Range and Polarity

Each device channel can be programmed for either unipolar or bipolar mode with gain settings of 1, 2, 4,
8, 16, 32, and 64. Note that the DaqBook/2020’s front panel analog inputs are bipolar only.
Channel Sampling Interval

The DaqBook/2000 Series and DaqOEM/2000 Series devices allow for programmable sampling intervalsof 5us, 10us, or 1000µs on a per channel basis. This mode allows some channels which change slowly but
a higher degree of accuracy is desirable to be sampled at a longer interval while channels that change more
rapidly to be sampled using a shorter interval. Each 5us or 10us interval reduces the maximum aggregateacquisition rate for the entire scan by that amount.
Note: DBK90 and DBK100 TC measurement modules sample at 1000µs/channel.

DaqBook/2000 Series and DaqOEM/2000 Series

898095

Device Overviews

2-3

Digital Input Channels

The devices allow for either synchronous scanning of digital input channels or asynchronous I/O
operations for all configured digital channels.

Counter Input Channels

The DaqBook/2000 Series and DaqOEM/2000 Series devices allow for synchronous scanning of the 4
16-bit counter input channels. The four 16-bit counter channels can also be cascaded into two 32-bit
counter channels. For either cascaded or non-cascaded counter channels each channel can be configured
for:
•
•

Pulse Counting Mode – specifies that each counter should be cleared upon being read and placed into
the input scan.
Totalize Counting Mode – specifies that each counter is to free-run and not be cleared during the input
acquisition.

Synchronous Input Acquisition Clocking

The DaqBook/2000 Series and DaqOEM/2000 Series devices allow clocking of the synchronized inputs
either by an internal, programmable pacer clock or by external clocking. These products use a sequencer
to implement a multiplexing approach to gathering the input data. This means that with either internal orexternal clocking the entire channel scan (including the sampling time for each channel) may not exceed
the maximum aggregate rate of 200 kHz.

Synchronous Output Operations
The DaqBook/2001, DaqBook/2020, and DaqOEM/2001 allow for synchronous output of D/A channels at
up to 100 kHz for each channel. All D/A channels can output waveform data from an on-board 256 k
sample buffer.
Output Channel Configuration
Analog Output Channels

For DaqBook/2001, DaqBook/2020, and DaqOEM/200, each D/A channel can be configured forwaveform output individually. If the D/A channel is not configured for waveform output it then isavailable for asynchronous output operations.

Digital Output Channel

May only be used for asynchronous digital I/O operations.

Synchronous Output Clocking

The DaqBook/2000 Series and DaqOEM/2000 Series devices allow clocking of the synchronized output by
the acquisition clock source, an internal, programmable pacer clock or by an external clock source. Whenthe clock source generates a new clock signal all outputs are updated concurrently. Regardless of the clock
source, the clock may not exceed the maximum update rate of 100 kHz.

Synchronous Output Data Source for DACs

For DaqBook/2001, DaqBook/2020, and DaqOEM/2001, the data source for synchronized output
operations can be that of the on-board 256k sample buffer.

2-4

Device Overviews

898095

DaqBook/2000 Series and DaqOEM/2000 SeriesAsynchronous I/O Operations
The DaqBook/2000 Series and DaqOEM/2000 Series devices allow asynchronous input of any counter ordigital channel that is not currently configured for synchronous acquisition. The devices also allow forasynchronous output to any D/A channels not currently configured for waveform output. Likewise, the16-bit P3 digital port, if available, can be used for both asynchronous input and output operations. In
addition, the timer outputs can be programmed at any time regardless of the current state of synchronous or
asynchronous operations on other channels.
Digital I/O Channels

Local 8255 Channels
The DaqBook/2000 Series and DaqOEM/2000 Series devices have an implemented Intel 8255 core in thedigital I/O logic on the P2 port of the product. With the Intel 8255 there are three 8-bit wide ports
available for I/O and one 8-bit wide port for configuration purposes. The configuration port is used toconfigure the other three 8-bit ports for either input or output operations.
Local 16-bit P3 PortThe 16-bit P3 Digital Port can be used as either an input, or an output port. With this port, noconfiguration is required, the port simply outputs when written to and inputs when read. The 16-bit P3
port does not apply to DaqBook/2020.
Expansion Digital I/O
The DaqBook/2000 Series and DaqOEM/2000 Series devices have the ability to expand digital I/O throughthe P2 port via digital I/O expansion modules. These modules are discussed in the DBK Option Cards &Modules User’s Manual. When using the digital I/O expansion modules the local P2 Intel 8255 digital I/O
becomes inaccessible in lieu of the expansion modules. These expansion modules provide additionallyIntel 8255 ports as well as input isolation for applications that require the expanded capabilities.
Pulse Stream Output Using Timers

The DaqBook/2000 Series and DaqOEM/2000 Series devices allow the generation of output pulses based
upon a programmable setting. These output timers can be set at any time regardless of the state of any
synchronous or asynchronous operations which are currently taking place on other channels.

Analog Output Channels

The DaqBook/2001, DaqOEM/2001, and DaqBook/2020 have the ability to output analog data to any of
the available (up to four) D/A channels. Each D/A channel may be asynchronously updated by anapplication if the D/A channel is not currently being used for waveform output operations. DaqBook/2005
and DaqOEM/2005 do not have analog output capability.

Counter Input Channels

The DaqBook/2000 Series and DaqOEM/2000 Series devices have counter input capabilities and have theability to read counter input [if the counter channel is not configured for synchronous acquisition]. As in
the case of synchronous operations the 4 16-bit counter input channels can be used individually or
cascaded into two 32-bit counter channels. For either cascaded or non-cascaded counter channels eachchannel can be configured for:•Clear on Read Mode - specifies that each counter should be cleared (reset to 0) upon being read.

•Continuous Totalize Mode – specifies that each counter is to free-run and not be cleared during
the read operation.

DaqBook/2000 Series and DaqOEM/2000 Series

898095

Device Overviews

2-5

Synchronous Operation Matrix
The following table applies to DaqBook/2000 Series and the DaqOEM/2000 Series devices. Note that thistable is closely related to the comparison table on page 2-1.
Operation

DaqBook/2020

DaqBook/2001
DaqOEM/2001

DaqBook/2005
DaqOEM/2005

Yes
Yes
Yes
Yes
Yes

Yes
Yes
Yes
Yes
Yes

Yes
Yes
Yes
Yes
Yes

Yes

Yes

No

Yes
Yes
Yes

Yes
Yes
Yes

Yes
Yes
Yes

Yes

Yes

No

Synchronous Input
Analog Main Unit Inputs
Analog Expansion Input
Counter Inputs
Digital Main Unit Inputs
Digital Expansion Inputs

Synchronous Output
Analog D/A Waveform Output

Asynchronous IO
Main Unit Digital I/O
Expansion Digital I/OTimer Output(Pulse Generation)
Analog Output

2-6

Device Overviews

898095

DaqBook/2000 Series and DaqOEM/2000 Series

DaqBook/2001 & DaqOEM/2001
16 Analog
Input
via P1

4 Analog
Outputvia P340 Digital
I/O
via P2 & P3

4 Counter In
2 Timer Out
via P3

Summary of DaqBook/2001 and DaqOEM/2001 Channels
DaqBook/2001 is a high-speed, multi-function, data acquisition module. DaqOEM/2001 is a board withthe same functionality, but no chassis. Both devices feature a 16-bit, 200-kHz A/D converter, digital
calibration, four 16-bit, 100-kHz D/A converters, 40 digital I/O lines, four counters, and two timers. Bothconnect to a PC or Ethernet network via a 10/100BaseT Ethernet port.
Up to 470 channels of analog and digital I/O can be accessed with one DaqBook/2001 or one
DaqOEM/2001 board. Up to 896 thermocouple channels can be accessed with the use of DBK90 and
DBK100 expansion modules.
P1, P2, and P3 37-pin connectors provide access to all I/O signals. Port descriptions follow:•

P1 – Analog input port for16 single-ended or 8 differential analog inputs with 13 software
programmable ranges (±10 V to ±156 mV full scale).

•

P2 – General purpose digital I/O port with 24 lines, or digital I/O expansion port controllingup to 192 external lines.

•

P3 – 16-bit digital I/O port, counter inputs, timer outputs, and analog outputs.

The on-board scan sequencer lets you select up to 16384 channel/range combinations. The sequencer scans
all channels of the scan at 5µs/channel, 10µs/channel, or 1000µs/channel.
100M-bit Ethernet allows analog and digital/counter input data, as well as analog and digital output data, toflow between the PC and the DaqBook/2001, or to the DaqOEM/2001.
DaqBook/2001 and DaqOEM/2001 each support a full complement of trigger modes including:•

Hardware analog triggering – A user-programmed trigger level sets an analog DAC, which iscompared in hardware to the analog input level on the selected channel. Trigger latency is < 5 µs.

•

Digital and pattern triggering – Each device has a separate digital trigger input line, allowing
TTL-level triggering and latencies less than 5 µs. The trigger can be programmed for logic levelor edge triggering. In pattern triggering, any of the digital input ports acts as the trigger port. You
can program the digital pattern.

•

Software-based triggering – The PC detects the trigger event from analog, digital, or counter
readings. Six pre- and post-triggering modes are supported. Trigger latency is one scan period.

The four 16-bit, 100-kHz analog output channels have an output from -10 V to +10 V.
Other features of the DaqBook/2001 and DaqOEM/2001 include:•

40 TTL-level digital I/O lines. They are divided into three 8-bit ports and one 16-bit port.

•

Four 16-bit counters. Each can accept frequency inputs up to 10 MHz. The counters can becascaded into two 32-bit counters.

•

Two 16-bit timer outputs. Each can generate square waves from 16 Hz to 1 MHz.

•

Configuration through software. There are no switches or jumpers on the DaqBook/2001 or on
the DaqOEM/2001.

DaqBook/2000 Series and DaqOEM/2000 Series

898095

Device Overviews

2-7

DaqBook/2005 & DaqOEM/2005
16 Analog
Input
via P1No DAC
Channels

40 Digital
I/O
via P2 & P3

4 Counter In
2 Timer Out
via P3

Summary of DaqBook/2005 and DaqOEM/2005 Channels
DaqBook/2005 is a high-speed, multi-function, data acquisition module. DaqOEM/2005 is a board withthe same functionality, but no chassis. Both devices feature a 16-bit, 200-kHz A/D converter, digital
calibra-tion, 40 digital I/O lines, four counters, and two timers. Both connect to a PC or Ethernet network
via a 10/100BaseT Ethernet port.
DaqBook/2005 and DaqOEM/2005 are identical to the DaqBook/2001 and DaqOEM/2001 previously
described, with one exception: the /2005 devices have no analog output capability.

DaqBook/2020
14 Analog
Input via P1
16 Differential
Voltage Inputs
via BNC
Connectors

14 Thermocouple
Inputsvia Mini-TC
Connectors

4 Analog
Outputvia BNC
Connectors

24 Digital
I/O via P2

4 Counter In
2 Timer Out
via BNC
Connectors

Note: DaqBook/2020’s Digital I/O is via the three 8-bit ports on
P2. The device has no P3 connector. For this reason it has
24 Digital I/O channels available, not 40 as do the otherdevices.

Note: In addition to the BNC connectors indicated above, the DaqBook/2020 includes a BNC for
External TTL Trigger input and one for External Clock input.

Summary of DaqBook/2020 Channels
DaqBook/2020 is a high-speed, multi-function, data acquisition module. It features a 16-bit, 200-kHz A/D
converter, digital calibration, four 16-bit, 100-kHz D/A converters, 24 digital I/O lines, four counters, andtwo timers. It connects to a PC or Ethernet network via a 10/100BaseT Ethernet port.
Up to 470 channels of analog and digital I/O can be accessed with one DaqBook/2020. Up to 896
thermocouple channels can be accessed with the use of DBK90 and DBK100 expansion modules.
P1, P2, and BNC connectors provide access to all I/O signals. Port descriptions follow:

2-8

Device Overviews•

P1 – Analog input port for14 single-ended analog inputs with 13 software programmable ranges(±10 V to ±156 mV full scale).

•

P2 – General purpose digital I/O port with 24 lines, or digital I/O expansion port controllingup to 256 external lines.

•

BNCs – DaqBook/2020 provides connection for 16 differential voltage inputs via BNC
connectors. In addition, BNC connectors are present for: 4 Counter Inputs, two Timer Outputs,
one External TTL Trigger Input, and one External Clock Input.

•

Mini-TC Connectors – DaqBook/2020 provides connection for 14 thermocouple inputs via
mini-TC connectors.
898095

DaqBook/2000 Series and DaqOEM/2000 Series

DaqBook/2020’s on-board scan sequencer lets you select up to 16384 channel/range combinations. The
sequencer scans all channels of the scan at 5µs/channel, 10µs/channel, or 1000µs/channel.
100M-bit Ethernet allows analog and digital/counter input data, as well as analog and digital output data, toflow between the PC and the DaqBook/2020.
DaqBook/2020 supports a full complement of trigger modes including:•

Hardware analog triggering – A user-programmed trigger level sets an analog DAC, which iscompared in hardware to the analog input level on the selected channel. Trigger latency is < 5 µs.

•

Digital and pattern triggering – DaqBook/2020 includes a digital trigger input line, whichallows TTL-level triggering and latencies less than 5 µs. The trigger can be programmed for logic
level or edge triggering. In pattern triggering, any of the digital input ports acts as the trigger port.
You can program the digital pattern.

•

Software-based triggering – The PC detects the trigger event from analog, digital, or counter
readings. Six pre- and post-triggering modes are supported.

The four 16-bit, 100-kHz analog output channels have an output from -10 V to +10 V.
Other features of the DaqBook/2020 include:•

24 TTL-level digital I/O lines. They are divided into three 8-bit digital I/O ports (on P2).

•

Four 16-bit counters. Each can accept frequency inputs up to 10 MHz. The counters can becascaded into two 32-bit counters.

•

Two 16-bit timer outputs. Each can generate square waves from 16 Hz to 1 MHz.

••

Configuration through software. There are no switches or jumpers on the DaqBook/2020.
16 differential voltage inputs via BNC connectors.

•

14 thermocouple inputs via mini-TC connectors.

DaqBook/2000 Series and DaqOEM/2000 Series

898095

Device Overviews

2-9

2-10

Device Overviews

898095

DaqBook/2000 Series and DaqOEM/2000 SeriesBlock Diagrams

3

DaqBook/2001, DaqOEM/2001, DaqBook/2005, & DaqOEM/2005 …… 3-1
DaqBook/2020 …… 3-2
Detail “TC” (Thermocouple Input Circuitry) for DaqBook/2020 …… 3-3
Detail “V” (Differential Voltage Input Circuitry) for DaqBook/2020 …… 3-3

DaqBook/2001, DaqOEM/2001, DaqBook/2005, & DaqOEM/2005

Note: The 4 channel analog output applies to
DaqBook/2001 and DaqOEM/2001. It
does not apply to DaqBook/2005 or to
DaqOEM/2005.

Block Diagram for use with
DaqBook/2001, DaqOEM/2001, DaqBook/2005, & DaqOEM/2005

DaqBook/2000 Series and DaqOEM/2000 Series

966895Block Diagrams

3-1

DaqBook/2020 Block Diagram

DaqBook/2020 Block Diagram

3-2

Block Diagrams

966895

DaqBook/2000 Series and DaqOEM/2000 Series

Detail “TC” (Thermocouple Input Circuitry) for DaqBook/2020

Detail “V” (Differential Voltage Input Circuitry) for DaqBook/2020

DaqBook/2000 Series and DaqOEM/2000 Series

966895Block Diagrams

3-3

3-4

Block Diagrams

966895

DaqBook/2000 Series and DaqOEM/2000 Series

Connectors, Indicators, Cables, & Pinouts

4

DaqBook/2020 …… 4-1
DaqBook/2001 and DaqBook/2005 …… 4-3
DaqOEM/2001 and DaqOEM/2005 …… 4-4
Cables …… 4-5
Pinouts …… 4-7

DaqBook/2020

DaqBook/2020
DaqBook/2020 provides for up to 256 channels of analog input expansion, 192 channels of digital
expansion I/O, and 896 channels of thermocouple measurement via expansion with DBK90 and DBK100
modules.
Front Panel
The front panel is populated with an upper row of mini-TC connectors and two rows of BNC connectors.
All connectors are clearly labeled on the device.
• Thermocouple Inputs – Mini TC connectors TC1 through TC7 and TC9 through TC15 provide ameans of connecting up to 14 thermocouples in total. Note that TC0 and TC8 are for cold
junction compensation.
• Analog Common – Ground jack used as a reference for differential measurements.
• Differential Voltage Inputs – 16 BNC connectors.
• Analog Output - 4 BNC connectors ( AO0, AO1, AO2, and AO3 ).
• Counters – Four BNC connectors (CTR0, CTR1, CTR2, and CTR3) for 16-bit counter input
signals.
• Timers – Two 16-bit Timer Output signals via BNC connectors (T0 and T1).
• Trigger – BNC for TTL triggering of an acquisition.
• External Clock – BNC for clocking each scan of an acquisition via an external clock.

DaqBook/2000 Series and DaqOEM/2000 Series

898095Connectors, Indicators, Cables, & Pinouts

4-1Rear Panel
The DaqBook/2020 rear panel contains P1 and P2 DB37 connectors, a MAC Address Label, DIN5 power
connector, power switch, 2 Synchronization Ports, Ethernet Port, and 5 LEDs [ 2 of which are on the
Ethernet connector].

DaqBook/2020, Rear Panel View
P1 - DB37 Connector: Analog input port for 14 single-ended analog inputs with 13 software
programmable ranges (±10 V to ±156 mV full scale). The Analog Input signals enter through P1 go through amultiplexer (MUX) to PGA, to Gain & Offset Amplifier, then to Analog-to-Digital Converter (ADC).

P2 - DB37 Connector: General purpose digital I/O port with 24 lines, or digital I/O expansion port
controlling up to 192 external lines.
MAC Address Label: The Media Access Control (MAC) label is located near the Ethernet connector. The
label shows the device serial number in three formats: (1) Barcode, (2) base 10, and (3) hexadecimal. If
prompted to enter a serial number in software, use the base 10 number. Conversion to a hexadecimal
number for use in addressing will be automatic.
ETHERNET: The 10/100BaseT Ethernet port can connect to the Ethernet port of the host PC, or to an
Ethernet network. Either of two Ethernet patch cables may be used to make the connection. CA-242 is a
1.5 foot cable. CA-242-7 is a 7-foot cable. Note that the Ethernet connector has two built in LEDs thatindicate traffic flow. These are discussed with the three other Ethernet-related LEDs. Note that the
Ethernet cable length must be <10m in order for the system to be CE Compliant.
LEDs: There are 5 ETHERNET Status LEDS. Two rectangular LEDs, Tx and Rx
are built right into the frame of the Ethernet jack. The other three, locatedjust to the right of the jack, are round and are labeled L, T, and D.
Tx – “ON” indicates traffic is being transmitted (see figure at right).
Rx – “ON” indicates that the port is receiving traffic.
L (Link) “ON” indicates a link exists. “OFF” indicates no link.
T (Transmit) “ON” indicates 100 M-Bits, “OFF” indicates 10 M-Bits.
D (Duplex) “ON” indicates full duplex, which allows simultaneous two-way data
traffic. “OFF” indicates half-duplex, which only allows one-way data
traffic at any given time.

Tx and Rx LEDs

SYNC (Qty of 2) – Two “synchronization ports” provide a means of synchronizing up to four
DaqBook/2000 Series units in regard to post-trigger scanning. The ports accept CA-74-1 andCA-74-5 RJ-11, 6-conductor type cables. Total combined cable length is not to exceed 15 feet (4.57 m).
POWER Switch: A rocker-type switch with a “0” label for Power Off, and a “1” for Power On.
POWER IN: +10 to +30 VDC, through a socket type DIN5 connector on the chassis. Power is typicallysupplied from a TR40U power adapter.

4-2

Connectors, Indicators, Cables, & Pinouts

898095

DaqBook/2000 Series and DaqOEM/2000 Series

DaqBook/2001 and DaqBook/2005
DaqBook/2001 and /2005 provide for up to 256 channels of analog input expansion. They have
40 built-in digital I/O channels, but can be expanded up to 272 channels.

DaqBook/2001
The lower unit is showing the rear panel.

The DaqBook/2001 and /2005 rear panels are identical. Each contains a P1, P2, and P3 DB37 connector,
a MAC Address Label, DIN5 power connector, power switch, 2 Synchronization Ports, Ethernet Port, and5 LEDs [ 2 of which are on the Ethernet connector].
P1 - DB37 Connector: Analog input port for 16 single-ended analog inputs or 8 differential,
programmable on a per-channel basis as single-ended or differential and as uni-polar or bipolar. The
13 programmable ranges are from ±10 V to ±156 mV full scale.
P2 - DB37 Connector: General purpose digital I/O port with 24 (3 x 8-bit) lines on P2, or digital I/O
expansion port controlling up to 272 external lines. Programmable as input or output.
P3 - DB37 Connector: 16-bit digital I/O port with 4 counter inputs and 2 timer outputs. In addition,
DaqBook/2001 has P3 connection to 4 analog outputs (DAC0, DA1, DAC2, and DAC3)..
MAC Address Label: The Media Access Control (MAC) label is located near the Ethernet connector. The
label shows the device serial number in three formats: (1) Barcode, (2) base 10, and (3) hexadecimal. If
prompted to enter a serial number in software, use the base 10 number. Conversion to a hexadecimal
number for use in addressing will be automatic.
ETHERNET: The 10/100BaseT Ethernet port can connect to the Ethernet port of the host PC, or to an
Ethernet network. Either of two Ethernet patch cables may be used to make the connection. CA-242 is a
1.5 foot cable. CA-242-7 is a 7-foot cable. Note that the Ethernet connector has two built in LEDs thatindicate traffic flow. These are discussed with the three other Ethernet-related LEDs. Note that the
Ethernet cable length must be <10m in order for the system to be CE Compliant.
LEDs: There are 5 ETHERNET Status LEDS. Two rectangular LEDs, Tx and Rx
are built right into the frame of the Ethernet jack. The other three, locatedjust to the right of the jack, are round and are labeled L, T, and D.
Tx – “ON” indicates traffic is being transmitted (see figure at right).
Rx – “ON” indicates that the port is receiving traffic.
L (Link) “ON” indicates a link exists. “OFF” indicates no link.
T (Transmit) “ON” indicates 100 M-Bits, “OFF” indicates 10 M-Bits.
D (Duplex) “ON” indicates full duplex, which allows simultaneous two-way data
traffic. “OFF” indicates half-duplex, which only allows one-way data
traffic at any given time.

Tx and Rx LEDs

SYNC (Qty of 2) – Two “synchronization ports” provide a means of synchronizing up to four
DaqBook/2000 Series units in regard to post-trigger scanning. The ports accept CA-74-1 and CA-74-5
RJ-11, 6-conductor type cables. Total combined cable length is not to exceed 15 feet (4.57 m).
DaqBook/2000 Series and DaqOEM/2000 Series

898095Connectors, Indicators, Cables, & Pinouts

4-3

POWER Switch: A rocker-type switch with a “0” label for Power Off, and a “1” for Power On.
POWER IN: +10 to +30 VDC, through a socket type DIN5 connector on the chassis. Power is typicallysupplied from a TR40U power adapter.

DaqOEM/2001 and DaqOEM/2005
The DaqOEM/2000 Series boards include all the analog, digital, and frequency I/O functionality that isfound in their DaqBook/2000 Series counterparts.
The DaqOEM/2000 Series boards are each powered by user-supplied +10 to +30 VDC input at 15W via a
DIN5 or 2-pin Molex connector. An optional AC/DC converter (TR-40U) can be used to supply theproper DC voltage from a standard 110 to 250 VAC source.

DaqOEM/2000 Series Board
All input and output signals for the DaqOEM/2000 Series boards are accessed via three 40-pin headersdesignated JP1, JP2, and JP3. Optional cables (p/n CA-248) are available for converting the 40-pin
headers to male DB37 connectors.
•

JP1 – 40 pin header. Analog input port for16 single-ended or 8 differential analog inputs
with 13 software programmable ranges (±10 V to ±156 mV full scale).

•

JP2 – 40 pin header. General purpose digital I/O port with 24 lines, or digital I/O
expansion port controlling up to 192 external lines.

•

JP3 – 40 pin header. 16-bit digital I/O port, 4 counter inputs, and 2 timer outputs.
In addition, for DaqOEM/2001 JP3 connects to 4 on-board analog outputs (DAC0, DAC1,
DAC2, and DAC3). DaqOEM/2005 has no on-board DACs.

•

TC Expansion – each device is expandable up to 896 TC inputs via use of DBK90 or
DBK100 options.

•

Analog Expansion – each device is expandable up to 256 analog input channels while
maintaining a 200 kHz scan rate.

In regard to specifications, for DaqOEM/2001 refer to DaqBook/2001 specifications. For
DaqOEM/2005 refer to DaqBook/2005 specifications.

4-4

Connectors, Indicators, Cables, & Pinouts

898095

DaqBook/2000 Series and DaqOEM/2000 Series

CablesShould you need help in regard to cables or have any other product inquiries please contact the factory.

Cables for use with DaqBook/2000 Series Devices
Description

Part No.

1.

Molded expansion cable. Connects a DBK
module to a DaqBook, 2 in.

CA-255-2T2.

Molded expansion cable. Connects a DBK
module to a DaqBook, 4 in.

CA-255-4T

3.

DBK expansion cable. Connects a DBK
module to a DaqBook, 2.5 in.

CA-37-1T

4.

DBK expansion cable. Connects a DBK
module to a DaqBook, 4.5 in.

CA-37-3T

5.

Expansion-card cable. Connects DBK
series expansion card options. The “x” inthe part no. indicates the number of cards
that can be connected via the cable.

CA-37-x

6.

SYNC cable, RJ-11, 6 conductor, 1 ft.

CA-74-1

7.

SYNC cable, RJ-11, 6 conductor, 5 ft.

CA-74-5

8.

Ethernet patch cable, shielded, 18 in.

CA-242

9.

Ethernet patch cable, shielded, 7 ft.

CA-242-7

10.

Power Cable, 5-pin male DIN to 5-pin
male DIN; provides convenient connection
between DaqBook power input connectors
and battery packs. 6 in.

CA-115

11.

Power Cable, 5-pin male DIN to vehicle
cigarette lighter, 8 ft.

CA-116

Cables for use with DaqOEM/2000 Series Boards
Description

Part No.

1.

SYNC cable, RJ-11, 6 conductor, 1 ft.

CA-74-1

2.

SYNC cable, RJ-11, 6 conductor, 5 ft.

CA-74-5

3.

Power Cable, 5-pin DIN to vehicle
cigarette lighter, 8 ft.

CA-116

4.

Ethernet patch cable, shielded, 18 in.

CA-242

5.

Ethernet patch cable, shielded, 7 ft.

CA-242-7

6.

Ribbon cable, 40-pin header to 37-pin
DSUB [DB37 connector], 9 in.

CA-248The total combined length of the SYNC cables is not toexceed 15 feet (4.57 m).

DaqBook/2000 Series and DaqOEM/2000 Series

898095Connectors, Indicators, Cables, & Pinouts

4-5

4-6

Connectors, Indicators, Cables, & Pinouts

898095

DaqBook/2000 Series and DaqOEM/2000 Series

Pinouts
P1 and JP1 – Analog I/O …… 4-8
P2 and JP2 – Digital I/O …… 4-9
P3 and JP3 – Pulse/Frequency/Digital I/O (Does not apply to DaqBook/2020E)…… 4-10

CAUTION
Turn off power to all devices connected to the system before connecting cables orsetting configuration jumpers and switches. Electrical shock or damage toequipment can result even under low-voltage conditions.

CAUTION
The discharge of static electricity can damage some electronic components.
Semiconductor devices are especially susceptible to ESD damage. You should
always handle components carefully, and you should never touch connector pins or
circuit components unless you are following ESD guidelines in an appropriate ESD
controlled area. Such guidelines include the use of properly grounded mats andwrist straps, ESD bags and cartons, and related procedures.

DaqBook/2001 and DaqBook/2005 each have three 37-pin connectors designated as P1, P2 and P3.
DaqOEM/2001 and DaqOEM/2005 have no DB37 connectors. Instead, these boards make use of three
40-pin headers designated as JP1, JP2, and JP3. A CA-248 ribbon cable can be used to obtain a DB37
connector for a 40-pin header. The cables are 9 inches long.
Signal assignment to connectors is as follows.
•

P1 and JP1 are used for Analog I/O•

P2 and JP2 are used for Digital I/O•

P3 and JP3 are used for Pulse/Frequency and Digital I/O

DaqBook/2020 has a P1 and P2 connector, but no P3. It does, however, have BNC connectors that are
directly related to certain P3 signals. The upcoming P3 pinout indicates when a signal applies to a BNC
connector on the DaqBook/2020.
To connect a DBK option to a DaqBook connect a CA-37-x cable to the appropriate DB37 connector[P1, P2, or P3] on the DaqBook and connect the free end of the cable to the DBK card or module. If you
are not familiar with DBK options, or need refreshed, review the DBK Basics section of the DBK Options
User’s Manual (p/n 457-0905); then review information regarding specific DBKs, as applicable. A PDF
version of the DBK manual is included on your data acquisition CD.

DaqBook/2000 and DaqOEM/2000 Series User’s Manual

937795Connectors, Indicators, Cables, & Pinouts

4-7

P1 and JP1

Analog I/O

P1, DB37 Connector for DaqBook/2000 Series Devices

JP1, 40-pin header for DaqOEM/2000 Series Boards

This notch isat the edge ofthe board.

P1 Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
----

Signal Name
+5 Volts-15 Volts
Expansion 7
Expansion 9
Expansion 5
Expansion 6
DGND
Negative Reference
Positive Reference
Not Connected
CH 15 (SE), or CH 7 LO DIFF
CH 14 (SE), or CH 6 LO DIFF
CH 13 (SE), or CH 5 LO DIFF
CH 12 (SE), or CH 4 LO DIFF
CH 11 (SE), or CH 3 LO DIFF
CH 10 (SE), or CH 2 LO DIFF
CH 9 (SE), or CH 1 LO DIFF
CH 8 (SE), or CH 0 LO DIFF
Signal Ground (SGND)
A/I Clock**+15 Volts
Expansion 8
Expansion 10
Not Connected
TTL Trigger**
Expansion 11
Not Connected
AGND
AGND
CH 7 (SE), or CH 7 HI DIFF
CH 6 (SE), or CH 6 HI DIFF
CH 5 (SE), or CH 5 HI DIFF
CH 4 (SE), or CH 4 HI DIFF
CH 3 (SE), or CH 3 HI DIFF
CH 2 (SE), or CH 2 HI DIFF
CH 1 (SE), or CH 1 HI DIFF
CH 0 (SE), or CH 0 HI DIFF
AGND
AGND
AGND

Description
Expansion +5 V power
Expansion -15 VDC power
Digital OUT, external ADDRESS select bit 3
Digital OUT, external ADDRESS select bit 1
Digital OUT, external GAIN select bit 1
Digital OUT, external GAIN select bit 0
Digital Ground
Analog, -5 V reference
Analog +5V reference
N/A
Ch 15 HI IN (single-ended mode) / Ch 7 LO IN (differential mode)
Ch 14 HI IN (single-ended mode) / Ch 6 LO IN (differential mode)
Ch 13 HI IN (single-ended mode) / Ch 5 LO IN (differential mode)
Ch 12 HI IN (single-ended mode) / Ch 4 LO IN (differential mode)
Ch 11 HI IN (single-ended mode) / Ch 3 LO IN (differential mode)
Ch 10 HI IN (single-ended mode) / Ch 2 LO IN (differential mode)
Ch 9 HI IN (single-ended mode) / Ch 1 LO IN (differential mode)
Ch 8 HI IN (single-ended mode) / Ch 0 LO IN (differential mode)
Sense Common (SGND)
External ADC Pacer Clock Input / Internal ADC Pacer Clock Output
Expansion +15 V power
Digital OUT, external ADDRESS select bit 2
Digital OUT, external ADDRESS select bit 0
N/A
Digital IN, External TTL Trigger Input
Digital OUT, Simultaneous sample and hold (SSH)
N/A
Analog Ground
Analog GroundCh 7 IN (single-ended mode) / Ch 7 HI IN (differential mode)
Ch 6 IN (single-ended mode) / Ch 6 HI IN (differential mode)
Ch 5 IN (single-ended mode) / Ch 5 HI IN (differential mode)
Ch 4 IN (single-ended mode) / Ch 4 HI IN (differential mode)
Ch 3 IN (single-ended mode) / Ch 3 HI IN (differential mode)
Ch 2 IN (single-ended mode) / Ch 2 HI IN (differential mode)
Ch 1 IN (single-ended mode) / Ch 1 HI IN (differential mode)
Ch 0 IN (single-ended mode) / Ch 0 HI IN (differential mode)
Analog Ground
Analog Ground
Analog Ground

JP1 Pins
1
3
5
7
9
11
13
15
17
19
21
23
25
27
29
31
33
35
37
2
4
6
8
10
12
14
16
18
20
22
24
26
28
30
32
34
36
38
39
40

** In addition to P1 pin 20 and pin 25, DaqBook/2020 has an “EXTCLOCK” BNC and a “TRIGGER” BNC.

4-8

Connectors, Indicators, Cables, & Pinouts

937795

DaqBook/2000 and DaqOEM/2000 Series User’s Manual

P2 and JP2

Digital I/O
JP2, 40-pin header for DaqOEM/2000 Series boards.

P2, DB37 Connector for DaqBook/2000 Series Devices

This side is the outer edgeof the board.
In regard to orientation,note that JP1 would
appear just below this JP2
header.

P2 Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
-------

Signal Name
Not Connected
Not Connected
Port B - B7
Port B - B6
Port B - B5
Port B - B4
Port B - B3
Port B - B2
Port B - B1
Port B - B0
DGND
Not Connected
DGND
Not Connected
DGND
Not Connected
DGND+ 5 Volt Supply
DGND+ 5 Volt Supply
DGND
Port C - C7
Port C - C6
Port C - C5
Port C - C4
Port C - C3
Port C - C2
Port C - C1
Port C - C0
Port A - A7
Port A - A6
Port A - A5
Port A - A4
Port A - A3
Port A - A2
Port A - A1
Port A - A0
DGND
DGND
DGND

Description
N/A
N/A
Digital I/O: P2 Digital Port B, Bit 7; or, P2 Expansion Address Bit 0 Out
Digital I/O: P2 Digital Port B, Bit 6; or, P2 Expansion Address Bit 1 Out
Digital I/O: P2 Digital Port B, Bit 5; or, P2 Expansion Address Bit 2 Out
Digital I/O: P2 Digital Port B, Bit 4; or, P2 Expansion Address Bit 3 Out
Digital I/O: P2 Digital Port B, Bit 3; or, P2 Expansion Address Bit 4 Out
Digital I/O: P2 Digital Port B, Bit 2; or, P2 Expansion RESET Output
Digital I/O: P2 Digital Port B, Bit 1; or, P2 Expansion WRITE Output
Digital I/O: P2 Digital Port B, Bit 0; or, P2 Expansion READ Output
Digital Ground
N/A
Digital Ground
N/A
Digital Ground
N/A
Digital Ground
Expansion +5 Volt Power
Digital Ground
Expansion +5 Volt Power
Digital Ground
Digital I/O: P2 Digital Port C, Bit 7; or, P2 Expansion Data Bit 7
Digital I/O: P2 Digital Port C, Bit 6; or, P2 Expansion Data Bit 6
Digital I/O: P2 Digital Port C, Bit 5; or, P2 Expansion Data Bit 5
Digital I/O: P2 Digital Port C, Bit 4; or, P2 Expansion Data Bit 4
Digital I/O: P2 Digital Port C, Bit 3; or, P2 Expansion Data Bit 3
Digital I/O: P2 Digital Port C, Bit 2; or, P2 Expansion Data Bit 2
Digital I/O: P2 Digital Port C, Bit 1; or, P2 Expansion Data Bit 1
Digital I/O: P2 Digital Port C, Bit 0; or, P2 Expansion Data Bit 0
Digital I/O: P2 Digital Port A, Bit 7
Digital I/O: P2 Digital Port A, Bit 6
Digital I/O: P2 Digital Port A, Bit 5
Digital I/O: P2 Digital Port A, Bit 4
Digital I/O: P2 Digital Port A, Bit 3
Digital I/O: P2 Digital Port A, Bit 2
Digital I/O: P2 Digital Port A, Bit 1
Digital I/O: P2 Digital Port A, Bit 0
Digital Ground
Digital Ground
Digital Ground

DaqBook/2000 and DaqOEM/2000 Series User’s Manual

937795

JP2 Pin
1
3
5
7
9
11
13
15
17
19
21
23
25
27
29
31
33
35
37
2
4
6
8
10
12
14
16
18
20
22
24
26
28
30
32
34
36
38
39
40

Connectors, Indicators, Cables, & Pinouts

4-9

P3 and JP3

Pulse/Frequency/Digital I/O
JP3, 40-pin header for DaqOEM/2000 Series boards.

This notch isat the edge ofthe board.

P3 PinSignal Name

Description

JP3 Pin

BNC /2020
Correlation

1
DGND
Digital Ground
1
2
DGND
Digital Ground3
3
Digital 7
P3 Digital Port Bit 7
5
4
Digital 6
P3 Digital Port Bit 6
7
5
Digital 5
P3 Digital Port Bit 5
96
Digital 4
P3 Digital Port Bit 4
11
7
Digital 3
P3 Digital Port Bit 3
13
8
Digital 2
P3 Digital Port Bit 2
15
9Digital 1
P3 Digital Port Bit 1
17
10
Digital 0
P3 Digital Port Bit 0
19
11
DGND
Digital Ground21
12
Expansion 2
Reserved for XC/D23
13
Expansion 3
Reserved for XWR25
14
Expansion 4
Reserved for XRD27
15
Timer 0P3 Timer 0 Output
29
16
Timer 1P3 Timer 1 Output
31
17Counter 2
P3 Counter 2 Input
33
18
Counter 0
P3 Counter 0 Input
35
19+ 15 Volts
Expansion, +15 VDC
3720+ 5 Volt Supply
Expansion, +5 Volt Power2
21
Analog Out Clock
External DAC Pacer Clock Input/Internal DAC Pacer Clock Output
4
22
Digital 15
Digital I/O; P3 Digital Port Bit 15
623
Digital 14
Digital I/O; P3 Digital Port Bit 14824
Digital 13
P3 Digital Port Bit 13
1025
Digital 12
P3 Digital Port Bit 121226
Digital 11
P3 Digital Port Bit 11
1427
Digital 10
P3 Digital Port Bit 10
1628
Digital 9
P3 Digital Port Bit 9
18
29
Digital 8
P3 Digital Port Bit 8
20
30
AGND
Analog Ground22
31
Analog Out DAC 3
Analog DAC 3 Output See note 1.
24
32
Analog Out DAC 2
Analog DAC 2 Output See note 1.
26
33
Analog Out DAC 1
Analog DAC 1 Output See note 1.
28
34
Analog Out DAC 0
Analog DAC 0 Output See note 1.
30
35Counter 3
P3 Counter 3 Input
32
36
Counter 1
P3 Counter 1 Input
34
37- 15 Volts
Expansion, - 15 VDC Power36--DGND
Digital Ground38--Reserved
Reserved for future use.
39--Reserved
Reserved for future use.
40
Note 1: Pins 31, 32, 33, and 34 connect to DAC0, DAC1, DAC2, and DAC3 for DaqBook/2001 and DaqOEM/2001.
DaqBook/2020 connects to the four DACs via BNC connectors. DaqBook/2005 and DaqOEM/2005 have no DACs.

4-10

Connectors, Indicators, Cables, & Pinouts

937795--------------BNC on 2020
BNC on 2020
BNC on 2020
BNC on 2020------------BNC on 2020
BNC on 2020
BNC on 2020
BNC on 2020
BNC on 2020
BNC on 2020-----

DaqBook/2000 and DaqOEM/2000 Series User’s Manual

Software

5

Overview …… 5-1
Out-of-the-BoxTM Software …… 5-1
Drivers for Third-party, Icon-driven Software …… 5-2
Language Drivers …… 5-2

Overview
The Daq devices have software options capable of handling most applications. Three types of software areavailable:• Out-of the-box graphical programs, e.g., DaqView, DaqViewXL, and post acquisition data
analysis programs such as eZ-PostView, eZ-TimeView, and eZ-FrequencyView.
• drivers for third-party, icon-driven software such as DASYLab and LabVIEW.
• various language drivers to aid custom programming using API; includes
DaqCOMTM ActiveX/COM support.

Out-of-theBoxTM Software
Out-of-the-Box programs are convenient for fill-in-the-blank applications that do not require programmingfor basic data acquisition and display:•

DaqView is a Windows-based program for basic set-up and data acquisition. DaqView lets you
select desired channels, gains, transducer types (including thermocouples), and a host of otherparameters with a click of a PC’s mouse. DaqView lets you stream data to disk and display data
in numerical or graphical formats.

•

DaqViewXL allows you to interface directly with Microsoft Excel to enhance data handling anddisplay. Within Excel you have a full-featured Daq control panel and all the data display
capabilities of Excel.

•

eZ-PostView provides a simple method of graphically viewing acquired waveforms from the
DaqBook Device. Up to 8 windows can be displayed on one screen, with up to 16 channels
overlaid on each window. eZ-PostView makes it simple to visually inspect acquired waveforms
from multiple channels within seconds of acquiring the data. Documentation, in Adobe PDF
format, is provided on the CD. Refer to the PostAcquisition Analysis.PDF.

•

eZ-TimeView & eZ-FrequencyView
eZ-TimeView and eZ-FrequencyView are optional post-acquisition analysis packages, which are
related to eZ-PostView, but include more features. eZ-TimeView is targeted at time-domain
analysis, including min/max, peak-peak, mean, RMS, plus a wide variety of plotting andwaveform viewing capabilities. eZ-FrequencyView is targeted at post-acquisition frequencydomain analysis, including FFT’s, octave analysis, plus dozens of other analysis features.
Documentation, in Adobe PDF format, is provided on the CD. Refer to the PostAcquisition
Analysis.PDF.

•

The Daq Configuration control panel allows for interface configuration, testing, andtroubleshooting.

DaqBook/2000 and DaqOEM/2000 Series User’s Manual

959295

Software5-1

Reference Notes:
¾
¾

PDF
Note:

The software document modules, DaqView, DaqViewXL, and Post Acquisition
Data Analysis User’s Guide, are not included as part of the hardcopy manual, but
are available in PDF version. See the PDF Note, below.
Programming topics are covered in the Programmer’s User Manual (1008-0901).
As a part of product support, this manual is automatically loaded onto your hard
drive during software installation. The default location is the Programs directory,
which can be accessed through the Windows Desktop.

During software installation, Adobe® PDF versions of user manuals will automaticallyinstall onto your hard drive as a part of product support. The default location is in the
Programs group, which can be accessed from the Windows Desktop. Refer to the PDF
documentation for details regarding both hardware and software.
Note that the PDF documents can be read directly from the CD by using the <View PDFs>
button located on the opening install screen.
A copy of the Adobe Acrobat Reader® is included on your CD. The Reader provides
a means of reading and printing the PDF documents. Note that hardcopy versions of themanuals can be ordered from the factory.

Drivers for Third-party, Icon-driven Software
LabVIEW® Support
We provide LabVIEW® support for expansion and signal conditioning modules. Our data acquisition VIs
for LabVIEW are more than just simple hardware access VIs, they are complete with engineering data
conversion, data display and logging capabilities.
DASYLab®
If your application requirements go beyond the scope of DaqView, you may want to consider DASYLab
software, as the latter offers a greater degree of flexibility and customizing.

Language Drivers
Each Daq system comes with an Application Programming Interface (API). API-language drivers include:
C/C++, and Visual Basic. The latest software is a 32-bit version API.
DaqBook/2000 Series devices are supplied with DaqX Subroutine API Libraries. These provide complete
support for all of the functionality available on each data acquisition device in Visual Basic®, and C/C++.
In addition, DaqX is supported under all versions of Windows®. Over 100 program examples are included
with DaqX Subroutine API Libraries. API documentation is provided in a Programmer’s Manual (p/n1008-0901), which is included on the installation CD.
The DaqCOM™ suite of programming allows applications developers to rapidly develop and deploy
custom systems by leveraging COM (Component Object Model) technology. DaqCOM does this byproviding a powerful easy-to-use interface to most programming languages including, Visual Basic®,
VBA, C++, and J++. In addition, DaqCOM supports the new Windows.NET architecture and includes
examples for VisualBasic.NET and C++. Support for VisualStudio.NET is accomplished via the
COMInterop feature within VisualStudio.NET.

5-2 Software

959295

DaqBook/2000 and DaqOEM/2000 Series User’s ManualConfiguring the System for Operation

6

Step 1 - Install the Software ….. 6-2
Step 2 - Determine the type of Network Connection …… 6-2
Step 3 - Connect System Components …… 6-5
Step 4 - Power-up the System …… 6-6
Step 5 - Configure the Computer’s Network Settings …… 6-7
Step 6 - Configure Device Network Settings using DaqIPConfig …… 6-10
Step 7 - Configure and Test the System using the Daq Configuration …… 6-11

Connecting a DaqBook/2000 Series Device to the Ethernet

The DaqBook/2000 and DaqOEM/2000 Series devices connect directly to an Ethernet port on a PC or
network hub via a built-in 10/100BaseT Ethernet interface. An Ethernet patch cable CA-242 (1.5 foot) or
CA-242-7 (7 foot) cable is used to make the connection.

Contact the factory or your service representative in regard to Ethernet connectivity ifyour operating system is other than Windows NT, Windows 2000, or Windows XP.

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-1

CAUTION
Turn off power to the system devices and externally connected equipment before connecting cables.
Electric shock or damage to equipment can result even under low-voltage conditions.
Take ESD precautions (packaging, proper handling, grounded wrist strap, etc.)

Reference Note: Adobe PDF versions of user manuals willautomatically install onto your hard drive as a part of productsupport. The default location is in the Programs group, which
can be accessed from the Windows Desktop. You can also
access documents directly from the data acquisition CD via the<View PDFs> button located on the CD’s opening screen.

Minimum System Requirements
PC system with Pentium® Processor
Windows Operating System
RAM as follows
32 Mbytes of RAM for Windows NT
64 Mbytes of RAM for Windows 2000
64 Mbytes of RAM for Windows XP

STEP 1 – Install the Software
1. Remove previous version Daq drivers, if present. You can do this through Microsoft’s Add/Remove
Programs feature.
2. Place the Data Acquisition CD into the CD-ROM drive. Wait for PC to auto-run the CD. This may take a
few moments, depending on your PC. If the CD does not auto-run, use the Desktop’s Start/Run/Browse
feature to locate and run Setup.exe [from the applicable CD-ROM drive].
3. After the intro-screen appears, follow the screen prompts.

STEP 2 – Determine the type of Network Connection
To properly connect and configure the device you must determine the type of network it will become partof. This is because the type of network used has a direct bearing on the IP address of the device.
Briefly, the four network types are as follows:• Dedicated Network - with a direct cable connection from the PC to the device• Dedicated Network - making use of a network hub or switch• LAN with a DHCP server(Local Area Network with a Dynamic Host Configuration Protocol)• LAN without a DHCP server(Local Area Network with no Dynamic Host Configuration Protocol)
Brief descriptions and illustrations follow.

6-2

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 Series

Dedicated Network - with a direct cable connection from the PC to the device
In this scenario a DaqBook/2020 is connected directly to an Ethernet jack on a host computer. The
following figure could just as easily portray a DaqBook/2001, DaqBook/2005, DaqOEM/2001, or
DaqOEM/2005 connected to the host PC via Ethernet cable.

Dedicated Network using a Direct Cable Connection
Dedicated Network - making use of a network hub or switch
In the following figure a DaqBook/2020 is connected to the Ethernet through a network hub/switch. At
least one computer is also connected to the hub. The other DaqBook/2000 Series and DaqOEM/2000
Series devices can be connected directly to the hub/switch in the same manner.

Dedicated Network using a Hub/Switch
Some network devices such as a wireless access point may act as a DHCP server. If this is
the case, follow the instructions for the LAN with a DHCP server. For detailed
information consult the documentation that is specific to your network device.

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-3

LAN with a DHCP Server (Local Area Network with a Dynamic Host Configuration Protocol server)
Many corporations use the LAN/Server with DHCP arrangement for their networks. In this type of setup
several computers are typically connected to a network that makes use of a DHCP server. In addition, theprimary data acquisition device, e.g., DaqBook/2020, is connected to the network hub/switch.

LAN with a DHCP Server
Notes:¾ Using a DaqBook/2020, DaqBook/2001, DaqBook/2005, DaqOEM/2001, or DaqOEM/2005 on a
typical LAN may affect the speed of the network and internet data transfer. Because of this werecommend adding a network card to the computer and using one of the two dedicated network
configurations.
¾

Contact your network administrator before connecting a DaqBook/2020, DaqBook/2001,
DaqBook/2005, DaqOEM/2001, or DaqOEM/2005 to a corporate network.

LAN with no DHCP Server(Local Area Network with no Dynamic Host Configuration Protocol server)
This scenario looks the same as that shown in the previous illustration, except there is no Dynamic Host
Configuration Protocol (DHCP). In this type of setup, one or more computers are connected to a network;and each computer has a static IP address.

6-4

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 Series

STEP 3 – Connect the System Components
Reference Note:
For information regarding connections, refer to Chapter 4, Connectors, Indicators, Cables, & Pinouts.

What you will need to connect a DaqBook/2000 or DaqOEM/2000 Series device to the Ethernet:• An available connection to the Ethernet. The connection can be either- an Ethernet jack on a computer or- an Ethernet jack on a hub that is connected to the Ethernet.
•

An Ethernet patch cable, e.g., a CA-242 (1.5 foot cable) or a CA-242-7 (7-foot cable).

1.

Connect the Ethernet cable to the Ethernet jack on the DaqBook/2000 or DaqOEM/2000 Series
device, as applicable.

2.

Connect the other end of the Ethernet cable to the Ethernet jack on the host computer or
network hub.

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-5

STEP 4 – Power-up the System
What you will need:
A power supply with a range of +10 VDC to +30 VDC.
The power supply needs to have a male DIN5 connector.
Note: The switching-mode power supply that is commonly used with these systems has an input
range of 100 VAC to 240 VAC at 50 Hz to 60 Hz. The power supply’s output [to thedevice] is typically 15 VDC @ 2.7 amps via a DIN5 connector.
Note: Various AC adapter models support power grids of USA, Europe, Japan, and Asia.
Reference Note: It is possible to use a VDC power source other than the commonlyused switching-mode power supply, often referred to as an adapter. However, be sure
to consult the power-related sections of DBK Basics before doing so. The DBK Basics
segment immediately follows chapter 1 of this user’s manual.
Reference Note:
Refer to the power-related sections of DBK Basics before powering up a system. The
DBK Basics section is located near the end of this user’s manual.

How to make the connection:
In the following steps we use the term “Daq Device” to mean a DaqBook/2020, DaqBook/2001,
DaqBook/2005, DaqOEM/2001, or a DaqOEM/2005, as applicable to your system.
1. Using the unit’s power switch, turn the Daq device “OFF.” The switch will be in the “0”position and the power LED will be unlit.
2. Connect the DIN5 end of the adapter’s cable to the power in connector on the Daq device.
3. Connect the adapter’s plug to a standard AC outlet.
4. If your adapter has a power switch, position it to “ON.”5. Turn ON the Daq device by placing the power switch to the “1” position. The power led willlight up.

6-6

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 Series

STEP 5 – Configure the Computer’s Network Settings [Applies to “dedicated networks” only]
If using a LAN (Local Area Network), which has a DHCP server, skip this section and continue
with STEP 7 - Configure and Test the System using the Daq*Configuration Applet (page 6-11).

If using a LAN (Local Area Network), which has no DHCP server, skip this section andcontinue with STEP 6 - Configure Device Network Settings using DaqIPConfig (page 6-10).

1.

Open the Control Panel by navigating from the Windows Desktop as follows:
Start Menu ⇒ Settings ⇒ Control Panel.

2.

Double-click the “Network and Dial-up Connections” icon.

3.

Double-click the “Network Connection” icon for the network that the DaqBook device is connected to.

Local Area Connection Status
4. In the “Local Area Connection Status” box (previous figure), click on the <Properties> button.
The “Local Area Connection Properties” box will appear (following figure).

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-7

Local Area Connection Properties

5.

Double-click the “Internet Protocol (TCP/IP)” component (previous figure).
The “Internet Protocol (TCP/IP) Properties” box will appear (following figure).

Internet Protocol (TCP/IP) Properties

6-8

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 SeriesConfigure the Computer’s TCP/IP settings as follows.

Internet Protocol (TCP/IP) Properties

6.

Select the “Use the following IP Address” radio button.

7.

Set the IP address field to 10.0.0.x where x is some number from 1 to 255.
Make sure that each computer on the dedicated network has a unique
IP address.

8.

Set the Subnet mask to 255.0.0.0. Note that the remaining fields can be left unchanged.

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-9

STEP 6 – Configure Device Network Settings using DaqIPConfig
Applies only to a LAN (Local Area Network), which has a no DHCP server.
If using a LAN (Local Area Network), which has a DHCP server, skip this section and continue
with STEP 7 - Configure and Test the System using the Daq*Configuration Applet (page 6-11).

The DaqIPConfig applet, designed for 32-bit Windows NT/2000/XP systems. DaqIPConfig allows you to change
the IP address of a device to match the address of a host computer. The applet is located in the program group forthe associated device and can be accessed from the Windows Desktop via the start menu.
1. Locate the DaqIPConfig Applet.
Locate the DaqIPConfig applet by navigating from the Windows’ Desktop as follows:
Start Menu ⇒ Programs ⇒ IOtech DaqX Software ⇒ DaqIPConfig
2. Open the DaqIPConfig Applet.
Click on the DaqIPConfig selection to open the applet.

DaqIPConfig
3. Select the device that is to have the address change.
Note: In the above figure there is only one device to select, i.e., DaqBook/2020.
4. Set the internet protocol (TCP/IP) settings to be compatible with host computer.
(a) Select the radio button labeled “Use the following IP address.”(b) Enter the new internet protocol settings. If needed, consult your network administrator for
acceptable numbers.
(c) Click the <OK> button.
5. Reboot the device.
The new IP address will not take affect until the device has been powered-off, then powered back on.
6.

Repeat steps 3, 4, and 5 for other devices in the system.
After configuring the network settings for all devices, proceed to Step 7.

6-10

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 Series

STEP 7 – Configure and test the System using the Daq* Configuration Applet
The Daq* Configuration applet, designed for 32-bit Windows NT/2000/XP systems, is located in the
Windows Control Panel. It allows you to add or remove a device and change configuration settings. The
included test utility provides feedback on the validity of current configuration settings, as well as
performance summaries.
1. Open the Daq* Configuration Applet.
a.

Open the Control Panel by navigating from the Windows’ Desktop as follows:
Start Menu ⇒ Settings ⇒ Control Panelb.

From the Control Panel, double-click the Daq* Configuration icon.

2. Add the first-level device to the list of installed devices.
The first-level device is the device that will be connected directly to the Ethernet, via a host computer’s
Ethernet jack or a jack on a network hub. The DaqBook/2020 is an example of a first-level device.
a.

Select the Computer image in the Device Inventory configuration tree (following figure).

b.

Click the <Add Device> button. The “Select Device Type” box will appear.

c.

Select the DaqBook/2020 from the list of devices.

d.

Click the <OK> button. The “Properties” box will appear for the selected device.

Using Daq* Configuration Device Inventory & Select Device Type to Add a Device
Note: Although the above figure indicates that a DaqBook/2020 was selected as the first level
device, we could similarly have selected another device, i.e., a DaqBook/2001 or a
DaqBook/2005.
If your device is a DaqOEM/2001 or a DaqOEM/2005 you will select DaqBook/2001 or
DaqBook/2005, respectively. This is because the DaqOEM/2000 Series boards areelectronically and functionally identical to their DaqBook/2000 Series counterparts.

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-11

3. Set the properties of the first-level device.
In this step you will set the device properties according to one of the following two methods, depending onwhether you have a “Dedicated Network” or a “LAN with DHCP Server Network.” Illustrations of the
network types are provided on page 6-3.
Users of Dedicated Networks follow these 2 steps.
a.

Enter the Serial Number of the first-level device, e.g., DaqBook/2020. In the following
screen shots the Serial Number is 800069.

b.

Select the “Auto IP Setting” radio button. Note that the IP Address of the DaqBook/2020
will automatically be calculated and displayed in the IP Address field as indicated in thefollowing left-hand figure.

c.

Click the <OK> button.

For DEDICATED Networks

For LAN with DHCP Server Networks

Daq* Configuration, Properties Dialog Boxes
Note: The above images are based on DaqBook/2020. We could similarly have selected another device.

Users of LAN with DHCP Server Networks follow these 3 steps.
The DaqIPConfig applet provides the Serial Number and the IP Address of the device. Users of LAN
with DHCP Server Networks will need to enter both numbers in the Daq* Configuration, Properties
dialog boxes (previous right-hand figure). Page 6-10 includes instructions for accessing DaqIPConfig.
If needed, refer to the upper right-hand figure in regard to radio-button and data entry locations.
a.

Enter the base 10 version of the Serial Number of the device, e.g., DaqBook/2020.

b.

Select the “Manual IP Setting” radio button.

c.

In the IP Address field, enter the IP address.

d.

Click the <OK> button.
¾ Provide your network administrator with the information located on the device’s
MAC label.
¾ Find out from the administrator if the IP Address will be changing. If so, see if you
can obtain a permanent IP Address that will be dedicated specifically to yourdevice.

6-12

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 Series

4. Test the system connections.
a.

Make sure the device has been properly installed and is powered-on.

b.

Make sure all cables are properly and securely connected.

c.

Click the “Test Hardware” tab.

d.

Click the <TCP/IP Test> button. This tests the Transmission Control Protocol / Internet
Protocol. The TCP/IP test has two components: Pinging Device and TCP Connection.
The Pinging Device test is an ICMP (Internet Control Message Protocol) ping test. In
addition to indicating either “Passed” or “Failed,” the test displays the round-trip time of theping, for example, <10 ms; and Packet Loss expressed as a percent of data lost.
A long ping round trip time, for example, >50 ms and/or any packet loss indicates a slow
network that is not optimized.
If the Pinging Device Test fails a <Details> button will appear just above the <Resource
Test> button. If this is the case, click the <Details> button to obtain possible causes andpossible solutions.
The result of the TCP (Transmission Control Protocol) test will be either passed or failed.
If the TCP test failed a <Details> button will appear just above the <Resource Test> button.
If this is the case, click the <Details> button to obtain possible causes and possible solutions.

e.

Upon completion of the TCP/IP test, click the <Resource Test> button. The Resource Testconsists of two components: Resource Tests and Performance Tests.
The Resource Test performs a quick communications check. If the host PC cancommunicate with the device a message that the test passed is displayed.
The Performance Tests obtain quantitative results using the current configuration of theacquisition device. The test results represent the maximum rates at which various operations
can be performed, and will vary according to port hardware capabilities.
The ADC FIFO Input Speed part of the test results in a display of the maximum rate at whichdata can be transferred from the tested device’s internal ADC FIFO to computer memory.
Results are given in samples/second, where a sample (2 bytes in length) represents a single
A/D value.
When testing a DaqBook/2000 or a DaqOEM/2000 Series device, if the unit does not
respond within 30 seconds perform the following steps:
1) reboot the system
2) upon power-up, re-open the Daq* Configuration applet
3) select another configuration setting4) reinitiate the test

This completes the procedure for connecting a DaqBook/2020, DaqBook/2001, DaqBook/2005,
DaqOEM/2001, or DaqOEM/2005 to the Ethernet.

DaqBook/2000 and DaqOEM/2000 Series

898095

Configuring the System for Operation

6-13

Reference Notes:
Depending on your application and system setup, related information can be found in thefollowing sources:¾¾

The DBK Options Manual (p/n 457-0905).
The Programmer’s Manual, (p/n 1008-0901).

Note: You can access PDF documents directly from the opening screen of the data acquisition CD
via the <View PDFs> button.

6-14

Configuring the System for Operation

898095

DaqBook/2000 and DaqOEM/2000 Series

CE Compliance & Noise Considerations

7

Overview ……7-1
CE Standards and Directives …… 7-1
Safety Conditions ……7-2
Emissions/Immunity Conditions ……7-2
CE Enhancements for DBKs .…… 7-3
CE Compliance for System Expansion …… 7-3
Noise Considerations …… 7-4

Overview
CE-compliant products bear the “CE” mark and include a Declaration of Conformity stating theparticular specifications and conditions that apply. The test records and supporting
documentation that validate the compliance are kept on file at the factory.
The European Union established CE standards in 1985. The standards include specifications for safety,
EMI emissions, and immunity from electromagnetic interference. Products that are intended for placementin the European Union must meet or exceed the standards and bear the "CE" mark.
Although not required in the USA, meeting or exceeding the CE standards is considered good engineering
practice, since doing so enhances safety while reducing noise and ESD problems.
In contracted and in-house testing, most Daq products met the required specifications. In many cases,
products that were not originally in compliance were redesigned accordingly. In noted instances, alternate
product versions, shield plates, edge guards, special connectors, or add-on kits are required to meet CE
compliance.

CE Standards and Directives
The electromagnetic compatibility (EMC) directives specify two basic requirements:
1.
2.

The device must not interfere with radio or telecommunications.
The device must be immune from electromagnetic interference from RF transmitters, etc.

The standards are published in the Official Journal of European Union under direction of CENELEC(European Committee for Electrotechnical Standardization). The specific standards relevant to Daq
equipment are listed on the product’s Declaration of Conformity.
The safety standard that applies to Daq products is EN 61010-1:1993 (Safety Requirements for Electrical
Equipment for Measurement, Control, and Laboratory Use, Part 1: General Requirements).
Environmental conditions include the following:•indoor use•

altitude up to 2000 m•

temperature 5°C to 40°C (41°F to 104°F)•

maximum relative humidity 80% for temperatures up to 31°C (87.8°F) decreasing linearly
to 50% relative humidity at 40°C (104°F)•

mains supply voltage fluctuations not to exceed ±10% of the nominal voltage•

other supply voltage fluctuations as stated by the manufacturer•transient overvoltage according to installation categories (overvoltage categories) I, II and III
For mains supply, the minimum and normal category is II•

pollution degree I or II in accordance with IEC 664

DaqBook/2000 and DaqOEM/2000 Series

897995

CE-Compliance & Noise Considerations

7-1

For clarification, terms used in some Declarations of Conformity include:•

pollution degree: any addition of foreign matter, solid, liquid or gaseous (ionized gases) that may
produce a reduction of dielectric strength or surface resistivity. Pollution Degree I has no influenceon safety and implies: the equipment is at operating temperature with non-condensing humidity
conditions; no conductive particles are permitted in the atmosphere; warm-up time is sufficient toavert any condensation or frost; no hazardous voltages are applied until completion of the warm-up
period. Pollution Degree II implies the expectation of occasional condensation.

•

overvoltage (installation) category: classification with limits for transient overvoltage, dependent
on the nominal line voltage to earth. Category I implies signals without high transient values.
Category II applies to typical mains power lines with some transients.

Safety Conditions
Users must comply with all relevant safety conditions in the user’s manual and the Declarations ofConformity. This manual and Daq hardware make use of the following Warning and Caution symbols:
If you see either of these symbols on a product, carefully read the related information and be alert to thepossibility of personal injury.
This warning symbol is used in this manual or on the equipment to warn of possible
injury or death from electrical shock under noted conditions.
This warning/caution symbol is used to warn of possible personal injury or equipment
damage under noted conditions.
Daq products contain no user-serviceable parts; refer all service to qualified personnel. The specificsafety conditions for CE compliance vary by product; but general safety conditions include:
•
•
•
•

•

The operator must observe all safety cautions and operating conditions specified in the
documentation for all hardware used.
The host computer and all connected equipment must be CE compliant.
All power must be off to the device and externally connected equipment before internal access to thedevice is permitted.
Isolation voltage ratings: do not exceed documented voltage limits for power and signal inputs.
All wire insulation and terminal blocks in the system must be rated for the isolation voltage in use.
Voltages above 30 Vrms or ±60 VDC must not be applied if any condensation has formed on thedevice.
Current and power use must not exceed specifications. Do not defeat fuses or other over-current
protection.

Emissions/Immunity Conditions
The specific immunity conditions for CE compliance vary by product; but general immunity conditions
include:• Cables must be shielded, braid-type with metal-shelled connectors. Input terminal connections are to
be made with shielded wire. The shield should be connected to the chassis ground with the hardware
provided.
• The host computer must be properly grounded.
• In low-level analog applications, some inaccuracy is to be expected when I/O leads are exposed to
RF fields or transients over 3 or 10 V/m as noted on the Declaration of Conformity.

7-2

CE-Compliance & Noise Considerations

897995

DaqBook/2000 and DaqOEM/2000 Series

CE Enhancements for DBKs
The following CE enhancements are detailed in the DBK Cards and Modules User’s Manual(p/n 457-0905).
• DBK41/CE• Edge Guard (for DBK5, DBK8, and DBK44)• Applicable cables and connectors

CE Compliance for System Expansion
The DaqBook/2000 and DaqOEM/2000 Series devices are CE compliant at the time they leave the factory,
and will remain in compliance providing that the conditions stated on their respective Declaration ofConformity cards continue to be met.
You may want to expand your data acquisition system while having it retain its “CE Compliant” status. If
so, refer to the most recent Declaration of Conformity for each device, prior to making any efforts to
expand the system.
Connecting a non-CE Compliant DBK module or card to your system will
make the system non-compliant.

In order to remain CE Compliant, DaqBook/2020, DaqBook/2001, and DaqBook/2005 require the use of
ferrite inductive collars and a ground line, which are included in an Ethernet CE Kit. Installation
instructions for the kit follow this chapter.
General rules of thumb regarding expansion, while remaining CE Compliant, are:
Review the most recent Declaration of Conformity for each device.
Use short cables.
When possible, use shielded cables.
When assembling or disassembling components, take ESD precautions, including the use ofgrounded wrist straps.
• Ensure that the host computer is CE Compliant.
• Ensure that expansion devices are CE Compliant• Ensure all system components are properly grounded.
••••

DaqBook/2000 and DaqOEM/2000 Series

897995

CE-Compliance & Noise Considerations

7-3

Noise Considerations
Controlling electrical noise is imperative because it can present problems even with the best measurement
equipment. Most laboratory and industrial environments suffer from multiple sources of electrical noise.
For example, AC power lines, heavy equipment (particularly if turned on and off frequently), local radio
stations, and electronic equipment can create noise in a multitude of frequency ranges.
Local radio stations are a source of high frequency noise, while computers and other electronic equipment
can create noise in all frequency ranges. Creating a completely noise-free environment for test andmeasurement is seldom practical. Fortunately, simple techniques such as using shielded/twisted pair wires,
filtering, and differential voltage measurement are available for controlling the noise in our measurements.
Some techniques prevent noise from entering the system; other techniques remove noise from the signal.
While many techniques for controlling noise in signals provide a means of removing the noise that isalready present, the preferred solution is to prevent the occurrence of noise in the signal in the first place.
The following practices, some of which are required for CE compliance, should be employed to minimize
noise.
•

Make a solid earth ground connection. Required for CE Compliance. Ensure that the chassis of the
primary data acquisition device, e.g., DaqBook/2020 is connected to earth ground. This practice:(a) keeps radiated emissions low by keeping the chassis electrically quiet, (b) keeps potential commonmode voltages low, (c) improves user safety, and (d) provides a safe path for Electrostatic Dischargeenergy back to earth ground.

•

Use short Ethernet cables. The use of short Ethernet cables will reduce noise. The shorter the cable
the better.

•

Use shielded cables. Loose wires are effective antennae for radio frequency pickup and can form
loops for inductive pickup. The use of properly connected shields will greatly reduce such noise.

•

Minimize ambient EMI. The lower the ambient EMI, the better. Sources of electromagnetic
interference include solenoids, motors, computer equipment, high power distribution wiring, etc.

•

Distance cables. Power supply switch transients can vary in strength and frequency. Ethernet cables
can radiate digital switching noise. For these reasons route the power supply and Ethernet cables suchthat they are as far as possible from all analog lines. The analog lines include those that connect to P1
on the front panel, as well as those that connect to the rear panel via an expansion port, or through theP4 connector.

•

Use ferrite inductive collars. A clamp-on ferrite collar can be secured at each end of the Ethernet
cable and at both ends of the power supply cable. The collars on the Ethernet cable will reduce digital
switching noise. Note that the collars will not reduce the integrity of the Ethernet channel if thesystem is making use of shielded Ethernet cables, part number CA-242. Ferrite collars on the power
cable will reduce the effects of power transients.

This partial view of the host PC shows the
Ethernet cable with a ferrite collar. The
PC’s connection to the ground-line is also
visible.

7-4

CE-Compliance & Noise Considerations

897995This partial view of a data acquisition deviceshows its connection to the ground-line. It also
shows two ferrite collars, the foremost of which isclamped to the power cable. The backgroundcollar is clamped onto the Ethernet cable.

DaqBook/2000 and DaqOEM/2000 SeriesEthernet CE Kit

Installation Instructions

Introduction

The Ethernet CE Kit includes three ferrite clamp-on inductive collars to reduce EMI. It also includes aground-line for safety. The kit is simple to install and consists of the following components.

• 3 Ferrite Inductive Collars with Tie-wrap, p/n L-8-1• 1 Ground-Line, p/n CA-250• 4 Washers, #8 External Tooth, p/n WA-5 – used to lock screws (2 washers per screw)• 1 Screw, #8-32x1/4 Phillips Pan-head, p/n HA-154-4 – used to secure the ground-line to athreaded splice-plate hole on the data acquisition device• 1 Screw, #6-32x5/16 Phillips Pan-head, p/n HA-2-5 – used to secure the ground-line to the PC

Connect the Ground Line
This ground line connection procedure is only required if you are not using a shielded
Ethernet cable [e.g., MCC’s CA-242] connected to an earth-grounded PC.

CAUTIONUse care when connecting the ground line to the acquisition device and the PC.
The use of screws which are too long can cause damage to circuitry.

1.

If using a desktop or tower PC, (a) remove a screw that secures the computer casing to thechassis. The first photo shows a good connection point for the PC ground. (b) Use the #6-32x5/16
screw and two #8 tooth washers, orientated as illustrated, to secure one end of the ground line to
the PC chassis.
If using a notebook (laptop) PC, connect one end of the ground-line to a known earth ground.
Depending on the ground chosen, you may need to use a screw other than the one supplied.

2.

On the data acquisition device, remove one of the four protective bumpers(protective ears).

3.

Using the #8-32x1/4 screw and two #8 tooth washers, orientated as
illustrated to the right, secure the free end of the ground line to the chassis of
the data acquisition device. Note that you will be making use of one of thethreaded holes which was used to secure the bumper. The second photo
below shows such a connection.

Ground Line on PC Chassis

Ground Line on Acquisition Device“Protective Bumper” shown removed.

4.

929195Re-attach the protective bumper to the acquisition device.

p/n1077-0901, rev 2.0

Install the Ferrite Inductive Collars
1. Place one ferrite collar at each end of the Ethernet cable.
Snap the two collars shut. See photos.
2. Place the remaining ferrite collar next to the DIN5 connector
on the power cable. Snap the collar shut.

3. Use a tie-wrap (one per ferrite collar) to secure each collar
to its respective cable.

Tie-wrap goes throughthis opening.

This partial view of the host PC shows the
Ethernet cable with a ferrite collar. The
PC’s connection to the ground-line is also
visible.

4. (Optional) Trim the excess tie-wrap material. Whiletrimming use care to avoid cutting into the cables.

This completes the installation.

1077-0901, rev 2.0

This partial view of the dataacquisition device shows its
connection to the ground-line. It also
shows two ferrite collars, the foremostof which is clamped to the power
cable. The background collar isclamped onto the Ethernet cable.

Ethernet CE Kit Instruction SheetTaking Measurements

8

Introduction …… 8-1
Scanning …… 8-1
Triggering …… 8-2
Synchronous I/O Operations…… 8-3
Asynchronous I/O Operations …… 8-7
Using DBK Cards and Modules for Signal Conditioning ….. 8-8

Introduction
With signal lines connected and a system up and running you can begin acquiring data. Specific connector
details and pinouts are provided in chapter 4. You should refer to the software documentation applicable toyour data acquisition system, for example, DaqView.

Scanning
DaqBook/2000 and DaqOEM/2000 Series devices each have an on-board scan sequencer that permits theuser to select any combination of up to 16384 channel/range combinations. The sequencer scans allchannels contained in the sequence at the rate of 5 µs/channel, 10 µs/channel, or 1000 µs/channel,
depending on the speed of the DBK expansion options; thereby minimizing the time-skew from channelto-channel. The user can also set the time between scan groups, from 0 to 6 hours. In addition to scanninganalog inputs, the sequencer can scan digital inputs and counter inputs.
Digital and frequency inputs can be scanned using the same scan sequencer employed for analog inputs,
enabling the time correlation of acquired digital data to acquired analog data.
(See Note)

5µs, 10µs, or 1000µs(See Note)

Scanning Example
Note:

All inputs can be measured at the 5 µs/channel rate. Some DBK expansion options require more settling time andcan be measured at the slower rate of 10 µs/channel or 1000 µs/channel. All types of measurements can bemixed within the scan group.

DaqBook/2000 and DaqOEM/2000 Series

898695

Taking Measurements

8-1

Triggering
Triggering can be the most critical aspect of acquiring data. DaqBook/2000 and DaqOEM/2000 Series
devices support a full complement of trigger modes to accommodate a wide variety of applications.
Multi-unit synchronization via the SYNC ports does not support pre-trigger data.
Only post-trigger data can be collected and time correlated between the units.

Hardware Analog Triggering. Many data acquisition products claim analog triggering, but rely on the
PC to take readings and make a decision, which leads to uncertain and potentially long latencies.
DaqBook/2000 and DaqOEM/2000 Series devices use true analog triggering, whereby the trigger level
programmed by the user, sets an analog DAC, which is then compared in hardware to the analog input
level on the selected channel. The result is analog trigger latency that is guaranteed to be less than 5 µs,
significantly shorter than most data acquisition devices. Any analog channel can be selected as the trigger
channel. The user can program both the trigger level, as well as the edge (rising or falling).
Digital Triggering (P1). A separate digital trigger input line is provided, allowing TTL-level triggering,again with latencies guaranteed to be less than 5 µs. Both the logic levels (1 or 0), as well as the edge
(rising or falling), can be programmed for the discrete digital trigger input. The digital trigger input islabeled TTLTRG.
Digital Pattern Triggering (P2 and P3 [if available]). The DaqBooks support digital pattern triggering,whereby the user can designate any of the digital input ports as the trigger port. The programmed digitalpattern, including the ability to mask or ignore specific bits, is then compared to the actual input until amatch is detected, after which the sequencer begins the scan sequence.
Counter Triggering. Triggering can be programmed to occur when one of the counters reaches, exceeds,
or is within a programmed level. Any of the built-in counter/totalizer channels can be programmed as a
trigger source.
Software-Based Triggering. Software-based triggering differs from the modes described above because
the readings [analog, digital, or counter] are interrogated by the PC to detect the trigger event, not in the
hardware as described above. The advantage of this mode is to permit triggering based on more complex
situations, such as on a specific temperature, which was derived from the acquisition of at least two analog
measurements, plus the calculation of the measured temperature using linearization algorithms.
Normally software-based triggering results in long latencies from the time that a trigger condition isdetected, until the actual capturing of data commences. However, the DaqBook circuitry circumvents thisundesirable phenomenon by use of pre-trigger data. Specifically, when software-based triggering is
employed, and the PC detects that a trigger condition has occurred, (which may be thousands of readings
later than the actual occurrence of the signal), the DaqBook driver automatically looks back to the location
in memory where the actual trigger-causing measurement occurred. The acquired data that is presented tothe user actually begins at that point. The latency in this mode is equal to one scan cycle.

8-2 Taking Measurements

898695

DaqBook/2000 and DaqOEM/2000 SeriesPre- and Post-Triggering Modes. Six modes of pre- and post-triggering are supported, providing a wide
variety of options to accommodate any measurement requirement. When using pre-trigger, the user mustuse software-based triggering to initiate an acquisition.

•

No pre-trigger, post-trigger stop event. This, the simplest of modes, acquires data uponreceipt of the trigger, and stops acquiring upon receipt of the stop-trigger event.

•Fixed pre-trigger with post-trigger stop event. In this mode, the user specifies the number of
pre-trigger readings to be acquired, after which, acquisition continues until a stop-trigger
event occurs.

•

No pre-trigger, infinite post-trigger. No pre-trigger data is acquired in this mode. Instead,data is acquired beginning with the trigger event, and is terminated when the operator issues acommand to halt the acquisition.

•

Fixed pre-trigger with infinite post-trigger. The user specifies the amount of pre-trigger datato acquire, after which the system continues to acquire data until the program issues acommand to halt acquisition.

•

Variable pre-trigger with post trigger stop event (driver support only). Unlike the previous
pre-trigger modes, this mode does not have to satisfy the pre-trigger number of readings
before recognizing the trigger event. Thus the number of pre-trigger readings acquired isvariable and dependent on the time of the trigger event relative to the start. In this mode, data
continues to be acquired until the stop trigger event is detected.

•

Variable pre-trigger with infinite post trigger (driver support only). This is similar to themode just described, except that the acquisition is terminated upon receipt of a commandfrom the program.

•

Stop Trigger. Any of the software trigger modes described above can be used to stop an
acquisition. Thus an acquisition can be programmed to begin on one event, such as atemperature level, and then stopped on another, such as a digital pattern event.

Synchronous I/O Operations
Synchronizing Multiple Units
DaqBook/2000 and DaqOEM/2000 Series devices can besynchronized to each other via their SYNC ports. Each
unit has two identical SYNC ports. Either, or both, SYNC
ports can be used to connect to the units via SYNC cables,
CA-74-1 or CA-74-5 (1-foot or 5-foot cables,
respectively). The units can be scan-synchronized andtriggered from any other SYNC-connected unit.
Synchronization Concept Block Diagram
The preceding diagram shows how a DaqBook can input or output synchronization timing and trigger
events on either SYNC connector. A more detailed block diagram follows shortly.
SYNC cables are not to exceed a total combined length of 15 feet (4.57 m).

Reference Note: The SYNC features are programmable via software. For detailed
information see the Using Multiple Devices and the daqAdcSetClockSource sections of
the Programmer’s Manual (p/n 1008-0901).
PDF NOTE:

During software installation, Adobe® PDF versions of user manuals automatically install onto
your hard drive as a part of product support. The default location is in the Programs group,
which can be accessed from the Windows Desktop. Refer to the PDF documentation for
details regarding both hardware and software.

DaqBook/2000 and DaqOEM/2000 Series

898695

Taking Measurements

8-3

DaqBook/2020 uses BNC
connectors for EXTCLK
and TRIGGER inputs.
The other devices use
pins as indicated in notes
1 and 2.

Synchronization Model for DaqBook/2000 and DaqOEM/2000 Series Devices,
Note 1: For DaqBook/2001 and DaqBook/2005, the clock signal is from P1, pin 20.
For DaqOEM/2000 Series devices, the clock signal is from JP1, pin 2.
Note 2: For DaqBook/2001 and DaqBook/2005 the trigger signal is from P1, pin 25.
For DaqOEM/2000 Series devices, the trigger signal is from JP1, pin 12.
In regard to synchronization, the following applies:•

One of three signals can be used to drive the DaqBook’s acquisition scan rate. These are:− the internal acquisition pacer clock− an external acquisition pacer clock.
− the global sync (G-SYNC) input from the SYNC ports•

Both the SYNC connector input and the external clock input can be divided down.

•

When a DaqBook is in the Master Mode, both the trigger condition and the scan timing are output
on the SYNC port. The global trigger (G-TRIG) is selected instead of the TTL Trigger Input.

Whether the DaqBook is using its internal scan clock, or external clock input, it can be programmed to
output the clock on the SYNC connector. In either case, the unit is behaving like a synchronization
master. Other DaqBook units [that are connected to the master, via a SYNC port] should be programmed
as synchronization slaves.
When a DaqBook is programmed as a synchronization slave, it will derive its scan period from the SYNC
port. If the slave unit must be triggered at the same time as the master unit, then the slave unit should use
TTL Trigger as its trigger source.
Multi-unit synchronization via the SYNC ports does not support pre-trigger data.
Only post-trigger data can be collected and time correlated between the units.

8-4 Taking Measurements

898695

DaqBook/2000 and DaqOEM/2000 Series

SYNC Notes:•

When a DaqBook is programmed as a synchronization slave, the TTL trigger source is
automatically derived from the SYNC port.

•

When a DaqBook is programmed as a synchronization master, it will output its trigger
condition on the SYNC port.

•

When synchronizing two or more DaqBooks, the slave units should have at least 0.1 µs ofdead time in the scan period. By “dead time,” we mean a duration in which no channels
are sampled. This accommodates fundamental differences in the clocks [for each
DaqBook].

•

All DaqBooks that are connected via SYNC cables can be scan-synchronized to within
0.1 µs of each other.

•

A maximum of 4 units can be synchronized; scan-synchronous (post trigger).

•

SYNC cables are not to exceed a total combined length of 15 feet (4.57 m).

Synchronous Input Operations
DaqBooks allow synchronous scanning and acquisition of Analog Input, Digital Input and Counter Input
Data at up to 200 kHz aggregate scanning rates. The Analog Input data can be either main unit or
expansion modules from P1 compatible analog input modules. The Digital Input data can be main unit 8bit P2 (8255) digital input or P2 compatible DBK digital input expansion modules. For DaqBook/2001,
DaqBook/2005 and the DaqOEM equivalents, 16-bit P3 digital inputs are applicable. DaqBook/2020 has
no P3 connector, but uses of BNC connectors for the same functionality.
Analog Input Channels
P1 – Channels 0 through 15

DaqBooks allow analog input configuration of themselves, as well as for the P1 compatible DBK analog
input expansion modules.
Channel Selection and Mode Settings

The DaqBook/2001, /2005 and DaqOEM/2001 and /2005 can accept up to 16 single ended or up to 8
differential-ended inputs and can be programmed for single-ended or differential-ended on a per channel
basis. Just one analog channel is sacrificed when a DBK expansion module is enabled. See DBK
documentation in the DBK Option Cards & Modules User’s Manual (p/n 457-0905) for additional
information.
DaqBook/2020 can accept up to 14 single-ended inputs. One analog channel is sacrificed when a DBK
expansion module is enabled. See DBK documentation in the DBK Option Cards & Modules User’s
Manual (p/n 457-0905) for additional information.
Channel Range and Polarity

DaqBook channels can be programmed for either unipolar or bipolar mode with gain settingsof 1, 2, 4, 8, 16, 32 and 64.
Channel Sampling Interval

DaqBooks allow for programmable sampling intervals of 5 us, 10 us, or 1000 us on a per channel basis.
This mode allows some DBK expansion channels [which change slowly but offer a higher degree of
accuracy] to be sampled at a longer interval, while channels that change more rapidly to be sampled using ashorter interval. Each 5 µs, 10 µs, or 1000 µs interval reduces the maximum aggregate acquisition rate forthe entire scan by that amount.

DaqBook/2000 and DaqOEM/2000 Series

898695

Taking Measurements

8-5Digital I/O Channels
There are 24 Digital I/O channels available to DaqBook/2020; and 40 available to DaqBook/2001, /2005,
and DaqOEM/2001 and /2005. The channels are grouped as follows:
8 Channels, A0 through A7 from P2 [or JP2] Port A
8 Channels, B0 through B7 from P2 [or JP2] Port B
8 Channels, C0 through C7 from P2 [or JP2] Port C
16 Channels, Digital 0 through Digital 15 from P3 [or JP3] These 16 channels do not apply to DaqBook/2020.

The DaqBooks allow for either synchronous scanning of digital input channels or asynchronous I/O
operations for all configured digital channels.
Counter Input Channels
Counter 0, 1, 2, 3 (CTR0, CTR1, CTR2, CTR3)

The DaqBooks allow synchronous scanning of four16-bit counter input channels. The 4 channels can becascaded into two 32-bit counter channels. For either cascaded or non-cascaded counter channels eachchannel can be configured for:••Pulse Counting Mode – specifies that each counter should be cleared upon being
read and placed into the input scan.
Totalize Counting Mode – specifies that each counter is to free-run and not becleared during the input acquisition.

Synchronous Input Acquisition Clocking
A /I Clock

The DaqBooks allow clocking of the synchronized inputs either by an internal, programmable pacer clock
or by external clocking. These products use a sequencer to implement a multiplexing approach togathering the input data. This means that with either internal or external clocking the entire channel scan(including the sampling time for each channel) may not exceed the maximum aggregate rate of 200 kHz.
Synchronous Output Operations for DaqBook/2020, DaqBook/2001, and DaqOEM/2001
DaqBook/2020, DaqBook/2001, and DaqOEM/2001 allow synchronous output of any D/A channels
available at up to 100 kHz for each channel. All D/A channels may have output clocked out
synchronously.
Output Channel Configuration
Analog Output Channels
DAC0, DAC1, DAC2, DAC3
(AO0, AO1, AO2, AO3)

Each D/A channel can be configured for waveform output individually. If the D/A channel is notconfigured for waveform output it then is available for asynchronous output operations.
Synchronous Output Clocking
A / O Clock

DaqBook/2020, DaqBook/2001, and DaqOEM/2001 allow clocking of the synchronized output by theacquisition clock source, an internal, programmable pacer clock or by an external clock source. When theclock source generates a new clock signal all outputs are updated concurrently. Regardless of the clock
source, the clock may not exceed the maximum update rate of 100 kHz.
Synchronous Output Data Source
DaqBook/2020, DaqBook/2001, and DaqOEM/2001 allow the data source for synchronized DAC output
operations to be that of an internal memory-based buffer. The output data, for all channels, is contained in
the buffer.

8-6 Taking Measurements

898695

DaqBook/2000 and DaqOEM/2000 SeriesAsynchronous I/O Operations
Each DaqBook allows asynchronous input of any counter or digital channel that is not currently configured
for synchronous acquisition. The circuitry allows for asynchronous output to any D/A channels that are
not currently configured for waveform output. Likewise, for DaqBook/2001, /2005 and DaqOEM/2001
and /2005 the 16-bit P3 [or JP3] digital port can be used for both asynchronous input and output
operations. In addition, the timer outputs can be programmed at any time regardless of the current state ofsynchronous or asynchronous operations on other channels.
Digital I/O Channels
Local 8255 Channels (P2 or JP2, Port A, Port B, Port C)
The DaqBooks have an implemented Intel 8255 core in the digital I/O logic on the P2 port. With the Intel
8255 there are three 8-bit wide ports available for I/O and one 8-bit wide port for configuration purposes.
The configuration port is used to configure the other three 8-bit ports for either input or output operations.
Local 16-bit P3 Port (P3 or JP3, Digital 0 through Digital 15) [Does not apply to DaqBook/2020]
The 16-bit P3 Digital Port can be used as either an input or output port. With this port, no configuration isrequired. The port simply outputs when written to and inputs when read.
Expansion Digital I/O
The DaqBooks have the ability to expand their digital I/O capabilities via the P2 port and the connection of
applicable digital I/O expansion modules. These modules are discussed in the DBK Option Cards &Modules User’s Manual (p/n 457-0905). When using the digital I/O expansion modules the local P2 Intel
8255 digital I/O becomes inaccessible in lieu of the expansion modules. These expansion modules provide
additional Intel 8255 ports as well as input isolation for applications that require the expanded capabilities.
Pulse Stream Output Using Timers (Timer0, Timer1)
Each DaqBook allows for the generation of output pulses based upon a programmable setting. These
output timers can be set at any time regardless of the state of any synchronous or asynchronous operations
which are currently taking place on other channels.
For DaqBook/2020, the outputs are through BNC connectors labeled T0 and T1.
For DaqBook/2001 and /2005, the timer outputs are through the P3 connector, pin 15 for Timer 0 and pin
16 for Timer 1.
For DaqOEM/2001 and /2005, the timer outputs are through the JP3 header, pin 29 for Timer 0 and pin 31
for Timer 1.
Analog Output Channels
DAC0, DAC1, DAC2, DAC3
(AO0, AO1, AO2, AO3)

DaqBook/2020, DaqBook/2001, and DaqOEM/2001 can output analog data to any of the four D/Achannels. Each D/A channel may be asynchronously updated by an application if the D/A channel is not
currently being used for waveform output operations.
For DaqBook/2020, the analog out channels make use of BNC connectors labeled:
AO0, AO1, AO2, and AO3.
For DaqBook/2001, the analog out channels use P3 as follows:
DAC0 – P3 pin 34, DAC1 – P3 pin 33, DAC2 – P3 pin 32, DAC 3 – P3 pin 31.
For DaqOEM/2001, the analog out channels use JP3 as follows: DAC0 – JP3 pin 30, DAC1 – JP3 pin 28,
DAC2 – JP3 pin 26, DAC 3 – JP3 pin 24.

DaqBook/2000 and DaqOEM/2000 Series

898695

Taking Measurements

8-7Counter Input Channels (CTR0, CTR1, CTR2, CTR3)
DaqBooks and DaqOEMs have counter input capabilities and have the ability to read counter input
providing that the counter channel is not configured for synchronous acquisition. As in the case ofsynchronous operations the four 16-bit counter input channels can be used individually or cascaded into
two 32-bit counter channels. For either cascaded or non-cascaded counter channels each channel can be
configured for:•Clear on Read Mode - specifies that each counter should be cleared (reset to 0) upon being read.

•Continuous Totalize Mode – specifies that each counter is to free-run and not be cleared during
the read operation.

Using DBK Cards and Modules for Signal Conditioning
The DBK signal-conditioning cards and module are designed for use with DaqBooks, LogBooks, andvarious types of data acquisition boards, i.e., ISA, PCI, and compact-PCI types. The DBKs perform bestwhen used with an acquisition device [such as a DaqBook] that can dynamically select channel, gain, andrange. DBK cards and modules with dynamic channel and gain/range selection allow for high channel-tochannel scan rates with a variety of transducers.
Note: Only passive DBKs, such as the DBK1 BNC module, the DBK11A screw terminal card, and the
DBK40 BNC analog interface, can be used with Daq PC-Cards.
DBK output signals can be bipolar, e.g., -5 to +5 V, or unipolar, e.g., 0 to 10 V. The user can select a
range of relevant values to correspond to the lowest signal (e.g., -5 or 0 V) and the highest signal (e.g., 5 or
10 V) signal. This type of range selection guarantees the highest resolution in 12-bit or 16-bit conversion.
DBK modules share the same footprint as DaqBooks and approximate the footprints of many notebook
PCs. The footprint factor allows for convenient stacking. The majority of these modules have their own
power supply; however, several options exist for packaging and powering the DBKs.
Reference Note:
DBK options are detailed in the DBK Option Cards and Modules User’s Manual(p/n 457-0905). As a part of product support, the manual is automatically loaded onto yourhard drive during software installation. The default location is the Programs group, which canbe accessed through the Windows Desktop. A PDF version of the document can be readdirectly from the data acquisition CD via the introduction screen’s <View PDFs> button.

8-8 Taking Measurements

898695

DaqBook/2000 and DaqOEM/2000 Series

Calibration

9

Overview …… 9-1
Using DaqCal.exe …… 9-1

Overview
The DaqBook/2000 and DaqOEM/2000 Series devices are digitally calibrated at the factory. The digital
method involves storing a correction factor on the circuit board for each range at the time of calibration.
Whenever a particular range is selected, the appropriate calibration constant is automatically applied to acompensating DAC, thereby calibrating the specific range. The result is that readings generated by the A/D
are already calibrated, and do not require additional processing.
This is significantly better than the calibration method in which readings are adjusted in software after they
are transferred to the host PC. That method reduces the dynamic range of the A/D, and can adversely
affect the speed at which the PC obtains a calibrated reading.
The DaqBook/2000 and DaqOEM/2000 Series devices include a user-cal mode, whereby the user canadjust the calibration of the unit’s board, without destroying the supplied factory calibration. This isaccomplished with two distinct calibration tables in the unit’s on-board EPROM, one which contains thefactory calibration factors, and the other which is available for user calibration.

Using DaqCal.exe
When calibration adjustments are needed they should be completed in the following order:
1.
2.
3.
4.
5.
6.
7.
8.

PGA Input and Output Offset
Sample/Hold Offset
A/D Offset and Gain
VDC Voltage Reference
DAC0 Full-Scale1
DAC1 Full-Scale1
DAC2 Full-Scale1
DAC3 Full-Scale1
1

DAC Full-Scale only applies to DaqBook/2020, DaBook/2001, and DaqOEM/2001.

A Windows-based program, DaqCal.exe, is used to calibrate Daq systems, including analog expansion
cards. DaqCal.exe is used in conjunction with:• a 4.5 digit, digital multi-meter• an adjustable voltage calibrator• an ambient temperature meter
To use the calibration program:1.

Launch DaqCal.
Note: DaqCal is installed automatically from your data acquisition CD as a part of productsupport. This takes place during software installation. DaqCal’s default location is the
IOtech DaqX Software folder, in the Programs group.

2.

When DaqCal opens you will be prompted to select your device from a list. After doing so,
simply follow the illustrated on-screen instructions.

DaqBook/2000 and DaqOEM/2000 Series

959095

Calibration

9-1

9-2

Calibration

959095

DaqBook/2000 and DaqOEM/2000 SeriesTroubleshooting and Customer Support

10

Electrostatic Discharge (ESD), Handling Notice…… 10-1
Product Care …… 10-1
ReadMe Files and the Install CD-ROM ……10-2
DaqCOM Issues …… 10-2
Driver Support……10-2
Ethernet Problems …… 10-3
Frequently Asked Questions …… 10-4
Customer Support …… 10-5

Electrostatic Discharge (ESD), Handling Notice
The discharge of static electricity can damage some electronic components. Semiconductor devices areespecially susceptible to ESD damage. You should always handle components carefully, and you should
never touch connector pins or circuit components unless you are following ESD guidelines in an
appropriate ESD-controlled area. Such guidelines include the use of properly grounded mats and wrist
straps, ESD bags and cartons, and related procedures.

Product Care
Daq products and DBK options are essentially maintenance free and require a minimal amount of care.
They should be treated much like any other high-tech equipment. In general:• Keep them clear of harsh chemicals and abrasive elements.
• Avoid exposing the products to extreme heat, for example, avoid setting up a DaqBook system near aboiler or furnace.
• Avoid extreme shock and vibration.
• Avoid subjecting the units to liquids and extremely fine air particulate, such as silica dust.
• Never touch circuit boards without proper ESD (Electrostatic Discharge) measures in place.
A “common-sense” approach to handling your data acquisition equipment will go a long way in protecting
it from inadvertent damage.
Note that you can use lint-free rags and Isopropyl Alcohol (Rubbing
Alcohol) to clean the outer plate surfaces of DaqBooks and optional
DBK modules.

DaqBook/2000 and DaqOEM/2000 Series – IO version

897995

Troubleshooting & Customer Support

10-1

ReadMe Files and the Install CD-ROM
The Install CD-ROM includes ReadMe Files. These files often contain late-breaking information that may
not appear in the user documentation. During installation you should review the ReadMe files when
prompted to by the program.
The Install CD-ROM includes:• DaqBook/2000 Series Windows NT/2000/XP driver• DaqX.DLL (32-bit API) for DaqBook/2000 Series• Microsoft C/C++ API support• Microsoft VB API support• Borland Delphi API support• Daq* Configuration Control Panel Applet• User documentation in Adobe® PDF format• DaqView
Upon installation, the user needs to run the Daq* Configuration Control Panel applet. The applet is used
to configure the DaqBook device for an appropriate Ethernet Port. This must be performed in order for theapplication to access the device through the Windows NT/2000/XP driver.

Driver Support
The daqX API can be used with WaveBook, DaqBook, DaqBoard, Daq PC Card and TempBook productlines. All daqX functions share the daq prefix.
The API support examples can be found in the installed directory:<Install> ProgrammingLanguageSupport\DAQ

Refer to the Programmer’s Manual, p/n 1008-0901.

DaqCOM Issues
DaqCOM support is contained within the Help file support, which is installed with DaqCOM. It is locatedwithin the DaqCOM program group.
DaqCOM support will not be available unless it has been specifically installed with the product. If the
DaqCOM program group is not listed under programs (which can be accessed via the Windows desktop),then you will need to install it from the data acquisition CD, or download it from our website.

10-2

Troubleshooting & Customer Support

897995

DaqBook/2000 and DaqOEM/2000 - IO version

Ethernet Problems• Make sure the Ethernet network is not experiencing technical problems. Check with your Network
Administrator.
• Ensure that your system is properly configured in the Daq*Configuration Applet [located in the
Windows Control Panel]. If a device is not configured correctly it will not be accessible from anapplication.
• Verify that the correct MAC (Media Access Control) number is reflected by the software.
• If communications cannot be established with the DaqBook or, if trying to connect causes the system
to hang or crash, try the following:-

Verify that the device power switch is set to ON and that the POWER ON indicator is lit.

-

Ensure that all physical device interface connections are correct. This includes Ethernet
connections from the DaqBook to the host PC’s Ethernet jack, or Ethernet Network hub.

-Run the Daq Configuration applet from the Windows Control Panel. Make sure the device
is configured to the appropriate Ethernet port with the appropriate protocol. Run the device
test and if it fails try other protocols for the device. Be sure to click on <Apply> to apply
the settings, before re-running the test.

• If problems persist, notify your Network Administrator. If he or she can’t solve the problem, contact
the factory.

DaqBook/2000 and DaqOEM/2000 Series – IO version

897995

Troubleshooting & Customer Support

10-3

Frequently Asked Questions(1) Topic: Environmental Factors
Question: What environments are the products intended for?
Answer:

Refer to the product specifications in regard to operating and storage temperature ranges, andrelative humidity. Note that system components are not intended to be exposed to harshenvironments and should always be protected from snow, rain, extreme dust, and harsh sun.

(2) Topic: DaqX
Question: What is DaqX software?
Answer:

DaqX software is low-level software support for application development. DaqX is installed
during the installation of the DaqView application. DaqX supports Visual C++ and Visual Basic
programming environments.

Question: Where can I get the latest version of DaqX software?
Answer:You can download the latest versions from our website. This service is free.

(3) Topic: Calibration
Question: Should I calibrate my DaqBook when I receive it?
Answer:No, because each DaqBook-related product is factory-calibrated to its rated accuracy before it
leaves the factory.

Question: When is calibration required?
Answer:Depending on your operating guidelines, your equipment will require periodic calibration toensure the original accuracy is maintained. We recommend that you have your DaqBook
system calibrated at least once a year.

(4) Topic: Computers for Data Acquisition
Question: What type of computer do you recommend for use with data acquisition devices such as a
DaqBook/2020E, DaqBook/2010E, or DaqBook/2010/OEM?
Answer:

10-4

The PC should have built in Ethernet capabilities and use an operating system of Windows NT,
2000, XP, or later.

Troubleshooting & Customer Support

897995

DaqBook/2000 and DaqOEM/2000 - IO version

Customer Support
If you want to Expand or Enhance Your DaqBook System . . .
You can visit our internet site atZZZPFFGDTFRP to find the latest accessories, expansion cards andmodules. You can also call our sales department directly at (440) 439-4091.

If you need to Report Problems or Request Product Support
Note:

Please do not return any equipment to the factory unless it has an RMA number (ReturnMerchandise Authorization number). RMA numbers are issued by the factory.

To report problems or request support, contact our applications department or your service representative.
Contact information follows shortly. When you contact us, please have the following information
available, as applicable:• Hardware model numbers and software version numbers.
•

Operating system, type of computer, and device information in the Windows control panel,such as interrupts and address settings for our hardware and others.

•

Results of tests, such as the Daq* Configuration control panel.

•

Hardware setup and software configuration.

•

Information on input signals, including voltage ranges, signal impedance ranges, noisecontent, and common mode voltages.

•

The name of a contact person from your company who can discuss the problems encountered.

••

Instructions for return shipping.
All troubleshooting notes and comments on tests performed, and all problem-related
conditions.
If your problem is Ethernet related, discuss the problem with your Network Administrator
prior to contacting the factory. Please provide us with your administrator’s comments, if
applicable.

•

Note: Before calling for assistance, take a few minutes to read all parts of the manual that may be relevantto the problem. Also, please review the troubleshooting material.
You can reach 0HDVXUHPHQW&RPSXWLQJ by one of the following means:

Knowledgebase: kb.mccdaq.com
Phone: 508-946-5100 and follow the instructions for reaching Tech Support
Fax: 508-946-9500 to the attention of Tech Support
Email: techsupport@mccdaq.com

Mail: 0HDVXUHPHQW&RPSXWLQJ&RUS. z &RPPHUFH:D\6XLWH

z 1RUWRQ, 0$All equipment returned to the manufacturer must have an RMA (Return Material Authorization) number.
You can obtain an RMA number by calling the Applications Department. When returning the equipment,
use the original shipping container (or equivalent) to prevent damage. Put the RMA number on yourshipping label to ensure that your shipment will be handled properly. After receiving your equipment, we
will fax a confirmation form that summarizes the charges (if applicable) and expected return date.

DaqBook/2000 and DaqOEM/2000 Series – IO version

897995

Troubleshooting & Customer Support

10-5

10-6

Troubleshooting & Customer Support

897995

DaqBook/2000 and DaqOEM/2000 - IO version

Specifications

11

DaqBook/2020, /2001, /2005, DaqOEM/2001, and DaqOEM/2005

GeneralSupply Voltage Range: 10 to 30 VDCPower Required: 15W, assuming no DBK options
Power Available for External Signal Conditioning & Expansion Options: 5V at 1A, ±15V at 500 mA, each not toexceed 7W on DaqBook/2020; not to exceed 10W on DaqBook/2001, /2005, DaqOEM/2001, or DaqOEM/2005
Operating Temperature: -30°C to + 70°C
Storage Temperature: -40°C to +80°C (-40° to 158°F)
Relative Humidity: 0 to 95% non-condensing
Vibration: MIL STD 810E, Categories 1 and 10
Signal I/O Connectors
DaqBook/2001 and /2005: DB37 male connectors: P1, P2, and P3
DaqBook/2020: DB37 male connectors: P1 and P2, BNC connectors, mini-Thermocouple Connectors
DaqOEM/2001 and /2005: 40-pin headers designated as JP1, JP2, and JP3. AMP # 2-103328-0
Dimensions
DaqBook/2001 and /2005: 285 mm W x 220 mm D x 45 mm H (11” x 8.5” x 1.75”)
DaqBook/2020: 285 mm W x 220 mm D x 90 mm H (11” x 8.5” x 3.5”)
DaqOEM/2001 and /2005: 27.5 cm W x 20.5 cm D
Weight
DaqBook/2001 and /2005: 1.36 kg (3 lbs)
DaqBook/2020: 2.5 kg (5.5 lbs.)
DaqOEM/2001 and /2005: 0.48 kg (1.06 lbs)

A/D Specifications
Type: Successive approximation
Resolution: 16 bit
Conversion Time: 5 µs
Maximum Sample Rate: 200 kHz
Nonlinearity (Integral): ±1 LSB
Nonlinearity (Differential): ± 2 LSB
Missing Codes: None, over full operating temperature range

Analog Inputs
Channels:
DaqBook/2001, 2005; via P1 (DB37): 16 single-ended or 8 differential, programmable on a per-channel basis as
single-ended or differential and as unipolar or bipolar.
DaqOEM/2001, /2005; via JP1 (40-pin header): 16 single-ended or 8 differential, programmable on a perchannel basis as single-ended or differential and as unipolar or bipolar.
DaqBook/2020; via BNC connectors: 16 differential inputs. via P1 (DB37): 14 single-ended.
Expansion: Up to 256 channels, without degradation, in maximum channel-to-channel scan rate
(5 or 10 µs per channel).
Total Harmonic Distortion: -84 dB (typ)Signal to Noise and Distortion: 80 dB (typ)Noise: 2 LSBRMS (typ)Bandwidth: 500 kHz

DaqBook/2000 and DaqOEM/2000 Series

899595Specifications

11-1

Specifications are subject to change without notice.

Settling Time: 5 µs maximum to 1 LSB for full-scale step
Temperature Coefficient: ±(10ppm +0.3 LSB)/°C outside the range of 0° to 30°C
Input Impedance: 10 MΩ for single-ended; 20 MΩ for differential
Bias Current: <1nA at 0° to 30°C
Common Mode Rejection: 86 dB, DC to 60 Hz for gains < 8; >100 dB for gains >16
Maximum Input Voltage: +11V relative to analog common
Over-Voltage Protection:
DaqBook/2001, DaqBook/2005: ±35V
DaqOEM/2001, DaqOEM/2005: ±35V
DaqBook/2020: ±25V
Crosstalk: -100 dB DC to 60 Hz; 86 dB @ 10 kHz
Ranges: Software programmable via sequencer on a per-channel basis.
Voltage Range(Note 1)

0 to +10 V
0 to +5 V
0 to +2.5 V
0 to +1.25 V
0 to +0.625 V
0 to +0.3125 V-10 to +10 V-5 to +5 V
-2.5 to +2.5 V-1.25 to +1.25 V-0.625 to +0.625 V-0.3125 to +0.3125 V-0.156 to +0.156 V

Accuracy (Note 2)
One Year, 0 to 35°C±(% reading + % range)

Input Noise (Note 3)(LSB rms)
10 Hz to 200 kHz

Absolute
0.015 + 0.005
0.015 + 0.005
0.015 + 0.005
0.015 + 0.008
0.015 + 0.008
0.015 + 0.008
0.015 + 0.005
0.015 + 0.005
0.015 + 0.005
0.015 + 0.005
0.015 + 0.008
0.015 + 0.008
0.020 + 0.008

Typical
1
1
1
2
2
2
1
1
1
2
2
3
3

Notes:
1) Specifications assume differential input single channel scan, 200 kHz scan rate, unfiltered.
2) Accuracy specification is exclusive of noise. Measurements were taken at P1.
3) Inputs shorted to Signal Ground (SGND). 8192 samples.

Analog Input Specifications for DaqBook/2020 front panel BNC Connectors
Inputs: 5 µs maximum to 1 LSB for full-scale step
Maximum Voltage Range: ±10V
Input Impedance: 20 MΩ
Accuracy:
For Gains 1, 2: ± [0.1% of reading + 0.005% of range]
For Gains 4, 8, 16, 32, 64: ± [0.1% of reading + 250uV]
Total Harmonic Distortion: -84 dB (typ)Signal to Noise and Distortion: 80 dB (typ)Noise: 2 LSBRMS (typ)Maximum Input Voltage (without damage): ±25 V
3 dB Bandwidth: 2.6 MHz
Temperature Coefficient: 10 ppm for every °C outside the range of 0° to 50°C

11-2

Specifications

958893

DaqBook/2000 and DaqOEM/2000 User’s ManualSpecifications are subject to change without notice.

Thermocouple Input Specifications for DaqBook/2020 Mini-TC Connectors
TC/mV Connectors: 14 Mini-TC connectors, oriented in 2 rows of 7
Inputs: 14 differential TC/mV inputs, open TC detection per channel
TC Types: J, K, T, E, S, R, B, N28, N14; x100 voltage
Input Impedance: 40 MΩ (differential); 20 MΩ (single-ended)
Input Voltage Range: ± 100 mV
Input Bandwidth: 4 kHz
Input Bias Current: 10 nA (typ)
CMRR: -100 db (typ)Maximum Working Voltage (Signal + Common Mode): ±10V
Over-Voltage Protection: ±40V
Voltage Accuracy: ±(0.2% of the reading + 50 µV)
Temperature Coefficient for Voltage Accuracy: 10ppm for every ˚C outside the range of 0° to 50°C
TC Accuracy: Valid for one year, 18° to 28°C, see the following table.
Temperature Coefficient for Type T: ±0.03°C per °C ambient outside the range of 18 to 28˚C
Minimum Resolution: 0.1˚C for all TC types

TC Accuracy at Measurement Temperature in °C (±°C)
Type

Min

Max

-100

0

100

300

500

700

900

1100

J

-200

760

0.8

0.7

0.7

0.8

0.9

0.9

—

—

1400
—

K

-200

1200

0.9

0.8

0.8

0.9

1.1

1.1

1.2

1.3

—

T

-200

400

0.9

0.8

0.8

0.8

—

—

—

—

—

E

-270

650

0.8

0.7

0.7

0.7

0.8

—

—

—

—

S

-50

1768

—

3.1

2.4

2.0

2.0

1.9

2.0

2.1

2.1

R

-50

1768

—

3.1

2.1

2.0

1.9

1.9

1.7

1.9

2.0

B

50

1780

—

—

—

4.9

3.2

2.8

2.4

2.3

2.0

N28

-270

400

1.2

0.9

0.9

0.9

—

—

—

—

—

N14

0

1300

—

0.9

0.9

0.9

1.1

1.1

1.2

1.3

—

Accuracy conditions:•••Exclusive of thermocouple errors
Exclusive of noise
VCM = 0

DaqBook/2000 and DaqOEM/2000 Series

898695Specifications

11-3

Specifications are subject to change without notice.

Input Sequencer
Analog, digital and counter inputs can be scanned synchronously based either on an internal programmable timer, oran external clock source. Analog and digital outputs can be synchronized to either of these clocks.
Scan Clock Sources: 2
1. Internal, programmable from 5 µs to 5.96 hours maximum in 1 µs steps
2. External, TTL level input up to 200 kHz maximum
Programmable parameters per scan: channel (random order), gain, unipolar or bipolar
Depth: 16,384 locations
On-board Channel to channel scan rate: 5 or 10 µs per channel, programmable
Expansion Channel Scan Rate: 5 or 10 µs per channel, programmable; 1 ms with DBK90 or DBK100

External Acquisition Scan Clock Input
Maximum Rate: 200 kHz
Clock Signal Range: 0V to +5V
Minimum Pulse Width: 50 ns high, 50 ns low

External Sync Ports
Number of External Sync Ports: 2, on rear panel
Maximum Number of Units to be Synchronized: 4 units, scan-synchronous (post trigger)
Maximum Length of Sync Cables: 15 feet (4.57 m), total for all cables

Triggering
Trigger Sources:6, individually selectable for starting and stopping an acquisition. Stop acquisition can occur on adifferent channel than start acquisition; stop acquisition can be triggered via modes 2, 4, 5, or 6 described below.
Pre-trigger is supported with fixed or variable pre-trigger periods.
1. Single-Channel Analog Hardware Trigger.
Any analog input channel, including any of the analog expansion channels, can be software programmed as
the analog trigger channel.
Input Signal Range: -10V to +10V max
Trigger Level: Programmable, 11-bit resolution
Hysteresis: Programmable, 11-bit resolution
Latency: 5 µs max
2. Single-Channel Analog Software Trigger.
Any analog input channel, including any of the analog expansion channels, can be selected as the software
trigger channel. If the trigger channel involves a calculation, such as with temperature, then the driver
automatically compensates for the delay required to calculate the reading, resulting in a maximum latency of
one scan period.
Input Signal Range: Anywhere within the range of the selected trigger channel
Trigger Level: Programmable, 16-bit resolution, including “window triggering”
Hysteresis: Programmable, 16-bit resolution
Latency: One scan period max.
3. Single-Channel Digital Trigger: A separate digital input is provided for digital triggering.
Input Signal Range: -15V to +15V max.
Trigger Level: TTL
Minimum Pulse Width: 50 ns high, 50 ns low
Latency: 5 µs max.
4. Digital Pattern Triggering: 8 or 16-bit pattern triggering on any digital input. Programmable for trigger on
equal, above, below, or within or outside of a window. Individual bits can be masked for “don’t care”
condition.
Latency: One scan period maximum.

11-4

Specifications

898695

DaqBook/2000 and DaqOEM/2000 User’s ManualSpecifications are subject to change without notice.

5. Counter/Totalizer Triggering: Counter/totalizer inputs can trigger an acquisition. User can select to trigger
on a frequency or on total counts that are equal, above, below, or within/outside of a window.
Latency: One scan period maximum.
6. Software Triggering: Trigger can be initiated under program control.

Analog Outputs
Applies to DaqBook/2020, DaqBook/2001, and DaqOEM/2001
Does not apply to DaqBook/2005 or to DaqOEM/2005
Four analog output channels are updated synchronously relative to scanned inputs, and clocked from either aninternal onboard clock, or an external clock source. Analog outputs can also be updated asynchronously,
independent of any other scanning in the system.
Channels: 4 DAC channels (DAC0, DAC1, DAC2, and DAC3)
Resolution: 16 bits
Data Buffer: 256Ksample
Output voltage range: ±10 V
Output current: ±10 mA
Offset error: ±0.0045 V maximum
Digital Feedthru: 50 mV when updated
Gain error: ±0.01%
Update rate: 100 kHz maximum, 1.5 Hz minimum (no minimum with external clock)
Settling Time: 10 µs maximum to 1 LSB for full-scale step
Clock Sources: 4 programmable
1. Onboard D/A clock, independent of scanning input clock
2. Onboard scanning input clock
3. External D/A input clock, independent of external scanning input clock
4. External scanning input clock

Digital I/O
Channels:
DaqBook/2001, /2005: 40 built-in, expandable to 272 with external DBK options
DaqOEM/2001, /2005: 40 built-in, expandable to 272 with external DBK options
DaqBook/2020: 24 built-in, expandable to 256 with external DBK options
Input Scanning Modes: 2 programmable
1. Asynchronous, under program control at any time relative to input scanning
2. Synchronous with input scanning
Ports:
DaqBook/2001, /2005: 3 x 8-bit (82C55 emulation), and 1 x 16-bit, Programmable as input or output.
DaqOEM/2001, /2005: 3 x 8-bit (82C55 emulation), and 1 x 16-bit. Programmable as input or output.
DaqBook/2020: 3 x 8-bit (82C55 emulation). Each port is programmable as input or output.
Input Characteristics: 100 Ω series, 20 pF to common
Input protection: ±8 kV ESD clamp diodes parallel
I/O levels: TTL
Sampling Rate: 200 kHz max
Update Rate: Asynchronous under program control
Output Characteristics: Output 12 mA per pin, 200 mA total continuous (per bank of 24 outputs)

DaqBook/2000 and DaqOEM/2000 Series

898695Specifications

11-5

Specifications are subject to change without notice.

Frequency/Pulse Counters
Counter inputs can be scanned synchronously along with analog and digital scanned inputs, based either oninternal programmable timer, or an external clock source. Bus mastering DMA provides CPU and systemindependent data transfers, insuring data acquisition performance irrespective of other system activities.
Counters can be configured to clear when read, or to totalize and clear under program control.
Channels: 4 x 16-bit, cascadable as 2 x 32-bit
Frequency Measurement Rate: 10 MHz maximum
Input Signal Range: -15 V to +15 V
Input Characteristics: 2.7 kΩ series in parallel with 20 pF to common and 10 kΩ to +5 V
Input protection: ±8 kV ESD clamp diodes parallel
Trigger Level: TTL
Minimum pulse width: 50 ns high, 50 ns low

Frequency/Pulse Generators
Channels: 2 x 16-bit
Output Waveform: Square wave
Output rate: 1 MHz base rate divided by 1 to 65535 (programmable)
High level output voltage: 2.0 V minimum @ -3.75 mA, 3.0 V minimum @ -2.5 mA
Low level output voltage: 0.4 V maximum @ 2.5 mA

11-6

Specifications

898695

DaqBook/2000 and DaqOEM/2000 User’s ManualSpecifications are subject to change without notice.

Accessories and Cables
Cables for use with DaqBook/2000 Series Devices
Description

Part No.

1.

Molded expansion cable. Connects a DBK
module to a DaqBook, 2 in.

CA-255-2T2.

Molded expansion cable. Connects a DBK
module to a DaqBook, 4 in.

CA-255-4T

3.

DBK expansion cable. Connects a DBK
module to a DaqBook, 2.5 in.

CA-37-1T

4.

DBK expansion cable. Connects a DBK
module to a DaqBook, 4.5 in.

CA-37-3T

5.

Expansion-card cable. Connects DBK
series expansion card options. The “x” inthe part no. indicates the number of cards
that can be connected via the cable.

CA-37-x

6.

SYNC cable, RJ-11, 6 conductor, 1 ft.

CA-74-1

7.

SYNC cable, RJ-11, 6 conductor, 5 ft.

CA-74-5

8.

Ethernet patch cable, shielded, 18 in.

CA-242

9.

Ethernet patch cable, shielded, 7 ft.

CA-242-7

10.

Power Cable, 5-pin male DIN to 5-pin male
DIN; provides convenient connection
between DaqBook power input connectors
and battery packs. 6 in.

CA-115

11.

Power Cable, 5-pin male DIN to vehicle
cigarette lighter, 8 ft.

CA-116

Cables for use with DaqOEM/2000 Series Boards
Description

Part No.

1.

SYNC cable, RJ-11, 6 conductor, 1 ft.

CA-74-1

2.

SYNC cable, RJ-11, 6 conductor, 5 ft.

CA-74-5

3.

Power Cable, 5-pin DIN to vehicle cigarette
lighter, 8 ft.

CA-116

4.

Ethernet patch cable, shielded, 18 in.

CA-242

5.

Ethernet patch cable, shielded, 7 ft.

CA-242-7

6.

Ribbon cable, 40-pin header to 37-pin
DSUB [DB37 connector], 9 in.

CA-248The total combined length of the SYNC cables is not toexceed 15 feet (4.57 m).

DaqBook/2000 and DaqOEM/2000 Series

898695Specifications

11-7

Specifications are subject to change without notice.

Manuals
Manual Set (Set #1121-0900). This is a hardcopy set of manuals that is available for purchase.
The set includes the following three manuals:
DaqBook/2000 and DaqOEM/2000 Series User’s Manual (p/n 1121-0901)
Programmer’s Manual for developing custom programs using API commands.
(p/n 1008-0901)
DBK Option Cards & Modules User’s Manual. (p/n 457-0905)

Software
DaqViewXL for seamless execution of DaqView from within Microsoft Excel’s tool palette.
DASYLab icon-based data acquisition graphics, control, and analysis software.
eZ-TimeView post-acquisition time-domain analysis software
eZ-FrequencyView post-acquisition frequency domain analysis software

11-8

Specifications

898695

DaqBook/2000 and DaqOEM/2000 User’s ManualGlossary

Acquisition

A collection of scans acquired at a specified rate as controlled by the sequencer.

AnalogA signal of varying voltage or current that communicates data.

Analog-to-Digital
Converter (ADC)

A circuit or device that converts analog values into digital values, such as binary bits, for use indigital computer processing.

API

Application Program Interface. The interface program within the Daq system’s driver that
includes function calls specific to Daq hardware and can be used with user-written programs(several languages supported).

BipolarA range of analog signals with positive and negative values (e.g., -5 to +5 V); see unipolar.

BufferBuffer refers to a circuit or device that allows a signal to pass through it, while providing
isolation, or another function, without altering the signal. Buffer usually refers to:
(a)

(b)
(c)

A device or circuit that allows for the temporary storage of data during data transfers. Suchstorage can compensate for differences in data flow rates. In a FIFO (First In - First Out)buffer, the data that is stored first is also the first data to leave the buffer.
A follower stage used to drive a number of gates without overloading the preceding stage.
An amplifier which accepts high source impedance input and results in low source
impedance output (effectively, an impedance buffer).

Buffer Amplifier

An amplifier used primarily to match two different impedance points, and isolate one stage froma succeeding stage in order to prevent an undesirable interaction between the two stages. (Alsosee, Buffer).

ChannelIn reference to Daq devices, channel simply refers to a single input, or output entity.
In a broader sense, an input channel is a signal path between the transducer at the point of
measurement and the data acquisition system. A channel can go through various stages(buffers, multiplexers, or signal conditioning amplifiers and filters). Input channels are
periodically sampled for readings.
An output channel from a device can be digital or analog. Outputs can vary in a programmed
way in response to an input channel signal.

Common mode

Common mode pertains to signals that are identical in amplitude and duration; also can be usedin reference to signal components.

Common mode
voltageCommon mode voltage refers to a voltage magnitude (referenced to a common point) that isshared by two or more signals. Example: referenced to common, Signal 1 is +5 VDC and
Signal 2 is +6 VDC. The common mode voltage for the two signals is +5.5 VDC [(5 + 6)/2].

CrosstalkAn undesired transfer of signals between systems or system components. Crosstalk causes
signal interference, more commonly referred to as noise.

DigitalA digital signal is one of discrete value, in contrast to a varying signal. Combinations of binary
digits (0s and 1s) represent digital data.

Glossary

959395G-1

Digital-to-Analog
Converter (DAC)

A circuit or device that converts digital values (binary bits), into analog signals.

DIP switch

A DIP switch is a group of miniature switches in a small Dual In-line Package (DIP). Typically,users set these switches to configure their particular application.

Differential mode

The differential mode measures a voltage between 2 signal lines for a single channel. (Also see
single-ended mode).

Differential mode
voltage

Differential mode voltage refers to a voltage difference between two signals that are referenced
to a common point. Example: Signal 1 is +5 VDC referenced to common. Signal 2 is +6 VDC
referenced to common.
If the +5 VDC signal is used as the reference, the differential mode voltage is +1 VDC(+ 6 VDC - +5 VDC = +1 VDC).
If the +6 VDC signal is used as the reference, the differential mode voltage is -1 VDC(+ 5 VDC - +6 VDC = -1 VDC).

ESDElectrostatic discharge (ESD) is the transfer of an electrostatic charge between bodies having
different electrostatic potentials. This transfer occurs during direct contact of the bodies, orwhen induced by an electrostatic field. ESD energy can damage an integrated circuit (IC).

Excitation

Some transducers [e.g. strain gages, thermistors, and resistance temperature detectors (RTDs)]
require a known voltage or current. Typically, the variation of this signal through thetransducer corresponds to the condition measured.

Gain

The degree to which an input signal is amplified (or attenuated) to allow greater accuracy andresolution; can be expressed as ×n or ±dB.

Isolation

The arrangement or operation of a circuit so that signals from another circuit or device do notaffect the isolated circuit.
In reference to Daq devices, isolation usually refers to a separation of the direct link between thesignal source and the analog-to-digital converter (ADC). Isolation is necessary whenmeasuring high common-mode voltage.

Linearization

Some transducers produce a voltage in linear proportion to the condition measured. Other
transducers (e.g., thermocouples) have a nonlinear response. To convert nonlinear signals into
accurate readings requires software to calibrate several points in the range used and theninterpolate values between these points.

Multiplexer (MUX)

A device that collects signals from several inputs and outputs them on a single channel.

Sample (reading)

The value of a signal on a channel at an instant in time. When triggered, the ADC reads thechannel and converts the sampled value into a 12- or 16-bit value.

Scan

A series of measurements across a pre-selected sequence of channels.

Sequencer

A programmable device that manages channels and channel-specific settings.

Simultaneous
Sample-and-Hold

An operation that gathers samples from multiple channels at the same instant and holds thesevalues until all are sequentially converted to digital values.

Single-ended mode

The single-ended mode measures a voltage between a signal line and a common reference that
may be shared with other channels. (Also see differential mode).

G-2

959395Glossary

TriggerAn event to start a scan or mark an instant during an acquisition. The event can be defined invarious ways; e.g., a TTL signal, a specified voltage level in a monitored channel, a button
manually or mechanically engaged, a software command, etc. Some applications may use preand post-triggers to gather data around an instant or based on signal counts.

TTL

Transistor-Transistor Logic (TTL) is a circuit in which a multiple-emitter transistor has replaced
the multiple diode cluster (of the diode-transistor logic circuit); typically used to communicate
logic signals at 5 V.

Unipolar

A range of analog signals that is always zero or positive (e.g., 0 to 10 V). Evaluating a signal in
the right range (unipolar or bipolar) allows greater resolution by using the full-range of thecorresponding digital value. See bipolar.

Glossary

959395G-3

G-4

959395Glossary