GP01157D/06/EN/02.22-00
71575135
2022-08-01

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Proline Promag 10
Electromagnetic flowmeter
HART

Services

Proline Promag 10 HART

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

5"System" menu . . . . . . . . . . . . . . . . . . . 132

1.1
1.2
1.3

4
4
4
4

5.1
5.2

1.4

Document function . . . . . . . . . . . . . . . . . . . . .
Target group . . . . . . . . . . . . . . . . . . . . . . . . . .
Using this document . . . . . . . . . . . . . . . . . . . .
1.3.1 Symbols . . . . . . . . . . . . . . . . . . . . . . .
1.3.2 Information on the document
structure . . . . . . . . . . . . . . . . . . . . . . .
1.3.3 Structure of a parameter description . .
Associated documentation . . . . . . . . . . . . . . . .

4
5
5

5.4
5.5

2

"Guidance" menu . . . . . . . . . . . . . . . . . . . 6

2.1"Commissioning" wizard . . . . . . . . . . . . . . . . . . 6
2.1.1 Device identification . . . . . . . . . . . . . . 9
2.1.2 System units . . . . . . . . . . . . . . . . . . . 10
2.1.3 Totalizer 1 to n . . . . . . . . . . . . . . . . . 14
2.1.4 Process parameters . . . . . . . . . . . . . . 16
2.1.5 Measurement conditions . . . . . . . . . . 17
2.1.6 Current output . . . . . . . . . . . . . . . . . 21
2.1.7 Pulse/frequency/switch output . . . . . 26
2.1.8 Display . . . . . . . . . . . . . . . . . . . . . . . 37
2.1.9 Date/time . . . . . . . . . . . . . . . . . . . . . 39

"Device management" submenu . . . . . . . . . . . 133"User management" submenu . . . . . . . . . . . . 135
5.2.1 "Define access code" wizard . . . . . . . . 137"Connectivity" submenu . . . . . . . . . . . . . . . . 138
5.3.1 "Bluetooth configuration" submenu . . 138" Date / Time" submenu . . . . . . . . . . . . . . . . 139"Information" submenu . . . . . . . . . . . . . . . . . 141
5.5.1 "Device" submenu . . . . . . . . . . . . . . . 141
5.5.2 "Sensor electronic module (ISEM)"
submenu . . . . . . . . . . . . . . . . . . . . . 144
5.5.3 "Display module" submenu . . . . . . . . 144"Display" submenu . . . . . . . . . . . . . . . . . . . . 146"Software configuration" submenu . . . . . . . . . 150

3"Diagnostics" menu . . . . . . . . . . . . . . . . 42

3.1
3.2
3.3
3.4
3.5

"Actual diagnostics" submenu . . . . . . . . . . . . . 43
"Diagnostic list" submenu . . . . . . . . . . . . . . . . 45
"Simulation" submenu . . . . . . . . . . . . . . . . . . . 48
"Heartbeat Technology" submenu . . . . . . . . . . 53
"Diagnostic settings" submenu . . . . . . . . . . . . . 54
3.5.1 "Properties" submenu . . . . . . . . . . . . . 54
3.5.2 "Diagnostic configuration" submenu . . 54

4

"Application" menu . . . . . . . . . . . . . . . . 71

4.1

"Measured values" submenu . . . . . . . . . . . . . . 71
4.1.1 "Totalizer" submenu . . . . . . . . . . . . . . 73
"System units" submenu . . . . . . . . . . . . . . . . . 75
"Totalizers" submenu . . . . . . . . . . . . . . . . . . . 80
4.3.1 "Totalizer handling" submenu . . . . . . . 80
4.3.2 "Totalizer 1 to n" submenu . . . . . . . . . 80
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . . 85
4.4.1 "Process parameters" submenu . . . . . . 85
4.4.2 "Low flow cut off" submenu . . . . . . . . 88
4.4.3 "Partially filled pipe detection"
submenu . . . . . . . . . . . . . . . . . . . . . . 89
4.4.4 "Sensor adjustment" submenu . . . . . . . 91
4.4.5 "Calibration" submenu . . . . . . . . . . . . 92
4.4.6 "Electrode cleaning cycle" submenu . . . 94
"Current output" submenu . . . . . . . . . . . . . . . . 96
"Pulse/frequency/switch output 1" submenu . 107"Communication" submenu . . . . . . . . . . . . . . 125
4.7.1 "HART configuration" submenu . . . . . 125
4.7.2 "Information" submenu . . . . . . . . . . 126
4.7.3 "Output" submenu . . . . . . . . . . . . . . 128

4.2
4.3
4.4

4.5
4.6
4.7

Endress+Hauser

5.3

5.6
5.7

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 151

3

About this document

Proline Promag 10 HART

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the operating
menus.
It is used to perform tasks that require detailed knowledge of the function of the device:• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Symbols

Types of information
Preferred procedures, processes or actions
Permitted procedures, processes or actions
Forbidden procedures, processes or actions
Additional information
Reference to documentationA

Reference to page
Reference to graphic

1.3.2

Information on the document structure

The parameters of all the operating menus and the commissioning wizard are described inthis document.
• Guidance menu with the Commissioning wizard (→  6), which guides the user
automatically through all the device parameters that are required for commissioning• Application menu (→  71)• Diagnostics menu (→  42)• System menu (→  132)

4

Endress+Hauser

Proline Promag 10 HART

About this document

1.3.3

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter nameWrite-protected parameter =Navigation

Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Selection

List of the individual options for the parameter• Option 1• Option 2

User entry

Input range for the parameter

User interface

Display value/data for the parameter

Additional information

Additional explanations (e.g. in examples):• On individual options• On display values/data• On the input range• On the parameter function

1.4

Associated documentation

Technical Information

Overview of the device with the most important technical data.

Operating Instructions

All the information that is required in the various phases of the life cycle ofthe device: from product identification, incoming acceptance and storage, tomounting, connection, operation and commissioning through totroubleshooting, maintenance and disposal as well as the technical data anddimensions.

Sensor Brief Operating Instructions

Incoming acceptance, transport, storage and mounting of the device.

Transmitter Brief Operating
Instructions

Electrical connection and commissioning of the device.

Description of Parameters

Detailed explanation of the menus and parameters.

Safety Instructions

Documents for the use of the device in hazardous areas.

Special Documentation

Documents with more detailed information on specific topics.

Installation Instructions

Installation of spare parts and accessories.

The related documentation is available online:
W@M Device Viewer

On the www.endress.com/deviceviewer website, enter the serial number ofthe device: nameplate

Endress+Hauser Operations App‣ Scan the Data Matrix code: nameplate‣ Enter the serial number of the device: nameplate

Endress+Hauser

5"Guidance" menu

Proline Promag 10 HART

2

"Guidance" menu

Main functions for use – from fast and safe commissioning to guided support during
operation.
Navigation

 Guidance

Guidance‣ Commissioning

2.1

→ 6

"Commissioning" wizard

Complete this wizard to commission the device. NOTE: If you exit the wizard beforehand,the changes you made will be saved. For this reason, the device may be in an undefinedstate! In this case, reset the device to the default settings.
Navigation

 Guidance → Commissioning‣ Commissioning

6

Device tag→ 9Serial number→ 9

Firmware version→ 9

Device name→ 9Volume flow unit

→  10Volume unit

→  11Mass flow unit

→  12Mass unit→  12

Density unit

→  13Temperature unit

→  13

Conductivity unit

→  13

Assign process variable→  14

Unit totalizer 1 to n

→  14

Totalizer operation mode→  15Endress+Hauser

Proline Promag 10 HART

Endress+Hauser"Guidance" menu

Failure mode→  16

Flow damping→  16

Low flow cut off→  18On value low flow cutoff→  19Off value low flow cutoff→  19Pressure shock suppression→  17

Empty pipe detection→  20Empty pipe adjust value→  20

Full pipe adjust value→  20

Process variable current output→  21

Current range output→  21

Lower range value output→  22Upper range value output→  23Damping current output→  24

Fixed current→  24

Failure behavior current output→  24

Failure current→  25

Operating mode→  26

Assign frequency output→  28Minimum frequency value→  28

Measuring value at minimum
frequency→  29Maximum frequency value→  29Measuring value at maximum
frequency→  29Failure mode→  307

"Guidance" menu

8

Proline Promag 10 HART

Failure frequency→  30Assign pulse output 1→  31

Pulse width→  31

Value per pulse→  32

Switch output function→  32

Assign diagnostic behavior→  33

Assign limit→  34

Switch-on value→  35

Switch-off value→  36

Switch-on delay→  36

Switch-off delay→  36

Assign status→  36

Failure mode→  37

Value 1 display→  37

Value 2 display→  38

Value 3 display→  38

Value 4 display→  39Display damping→  39Time format→  39Time zone→  40Set date/time→  41

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

2.1.1
Navigation

Device identification
 Guidance → Commissioning

Device tagNavigation

 Guidance → Commissioning → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)Serial number
Navigation

 Guidance → Commissioning → Serial number

Description

Displays the serial number of the measuring device. The serial number can be used to
identify the measuring device and to retrieve further information on the measuring device,
such as the related documentation, via the Device Viewer or Operations app.
Additional information:
The serial number can also be found on the nameplate of the sensor and transmitter.

User interface

Character string comprising numbers, letters and special characters

Firmware version
Navigation

 Guidance → Commissioning → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

Device name
Navigation

 Guidance → Commissioning → Device name

Description

Displays the name of the transmitter.
Additional information:
The name can also be found on the transmitter's nameplate.

Endress+Hauser

9"Guidance" menu

User interface

Proline Promag 10 HART

Character string comprising numbers, letters and special characters

2.1.2
Navigation

System units Guidance → Commissioning

Volume flow unitNavigation

 Guidance → Commissioning → Volume flow unit

Description

Select volume flow unit.

10

Endress+Hauser

Proline Promag 10 HART

Selection"Guidance" menu

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dUS units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• MMft³/s• MMft³/min• MMft³/h• Mft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)Volume unitNavigation

 Guidance → Commissioning → Volume unit

Description

Select volume unit.

Endress+Hauser

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)11

"Guidance" menu

Selection

Proline Promag 10 HART

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega

US units• af• ft³• Mft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;beer)• bbl (us;tank)

Mass flow unitNavigation

 Guidance → Commissioning → Mass flow unit

Description

Select mass flow unit.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

Mass unitNavigation

 Guidance → Commissioning → Mass unit

Description

Select mass unit.

Selection

SI units
•g
• kg
•t

12

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

US units• oz• lb• STonEndress+Hauser

Proline Promag 10 HART

"Guidance" menu

Density unitNavigation

 Guidance → Commissioning → Density unit

Description

Select density unit.

Selection

SI units• g/cm³• g/m³• kg/l• kg/dm³• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

US units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)Temperature unitNavigation

 Guidance → Commissioning → Temperature unit

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Select temperature unit.

Selection

SI units• °C•K

US units• °F• °R

Conductivity unitNavigation

 Guidance → Commissioning → Conductiv. unit

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Endress+Hauser

Select conductivity unit.

13

"Guidance" menu

Selection

Proline Promag 10 HART

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/m
• kS/m• MS/m2.1.3

Totalizer 1 to n

Navigation

 Guidance → Commissioning

Assign process variable

Navigation

 Guidance → Commissioning → Assign variable

Description

Select process variable for totalizer.
Additional information:
If the option selected is changed, the device resets the totalizer to "0".

Selection• Off• Volume flow
• Mass flow

Unit totalizer 1 to n

Navigation

 Guidance → Commissioning → Unit totalizer 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select process variable totalizer unit.

Selection

SI units• g*• kg *• t**

US units• oz *• lb *• STon *Visibility depends on order options or device settingsor

14

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

SI units• cm³ *
• dm³ *
• m³ *• ml *• l*• hl *• Ml Mega **US units• af *• ft³ *• Mft³ *• fl oz (us) *• gal (us) *• kgal (us) *• Mgal (us) *• bbl (us;liq.) *• bbl (us;beer) *• bbl (us;oil) *• bbl (us;tank) *Imperial units• gal (imp) *• Mgal (imp) *• bbl (imp;beer) *• bbl (imp;oil) *Visibility depends on order options or device settingsorOther units
None **

Additional information

Visibility depends on order options or device settings

Description
The unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  75).
Selection
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  14).

Totalizer operation modeNavigation

 Guidance → Commissioning → Operation mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select totalizer calculation mode.

Selection• Net flow total• Forward flow total• Reverse flow total

Additional information

Selection• Net flow total option
The flow values in the forward and reverse flow directions are totalized and netted
against each other. Net flow is recorded in the flow direction.
• Forward flow total option
Only the flow in the forward flow direction is totalized.
• Reverse flow total option
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Endress+Hauser

15

"Guidance" menu

Proline Promag 10 HART

Failure modeNavigation

 Guidance → Commissioning → Failure mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify how the totalizer should behave in the event of a device alarm.
Additional information:
The failsafe mode that applies to any other totalizers or outputs is specified separately inother parameters and is not impacted by this setting.

Selection• Stop• Actual value• Last valid value

Additional information

Selection• Stop option
The totalizer is stopped in the event of a device alarm.
• Actual value option
The totalizer continues to totalize based on the current value measured; the device alarmis ignored.
• Last valid value option
The totalizer continues to totalize based on the last valid value measured before thedevice alarm occurred.

2.1.4
Navigation

Process parameters Guidance → Commissioning

Flow damping

Navigation

 Guidance → Commissioning → Flow damping

Description

Enter value for damping of the flow measured value in order to reduce the variability of
the flow measured value when exposed to interference.
Additional information:
The depth of the flow filter is determined by this setting. As the filter depth increases, sodoes the reaction time of the device.
- Value = 0: No damping. Damping of 0 is not recommended, as the measuring signal isthen so noisy that it is almost impossible to perform a measurement.
- Value > 0: Damping increases
Optimal damping depends on the measuring period.
Damping impacts the following measuring device variables:- Outputs- Low flow cut off- Totalizers

16

Endress+Hauser

Proline Promag 10 HART

User entry"Guidance" menu

0 to 15

2.1.5

Measurement conditions

Pressure shock suppression
Navigation

 Guidance → Commissioning

Pressure shock suppression

Navigation

 Guidance → Commissioning → Pres. shock sup.

Description

Enter time frame for signal suppression (= pressure shock suppression active), for example
to prevent the device from registering flow movements in the pipe when a valve is closed.
Additional information:
Pressure schock suppression is activated when the flow rate drops below the on value forlow flow cutoff. Output values when pressure shock suppression is active:- Flow: 0- Totalizer: Last valid value
Pressure shock suppression is deactivated when:- The time frame specified has elapsed and- Flow exceeds the off value for low flow cutoff

User entry

0 to 100 s

Additional information

ExampleWhen a valve is closed, momentarily strong fluid movements may occur in the pipeline,
which are registered by the device. These totalized flow values lead to a false totalizer
status, particularly during batching processes.

Endress+Hauser

17

"Guidance" menu

Proline Promag 10 HART

C

D

Q

2

1

D

3

4

6
5

t

A

B
A0012888

Q
t
A
B
C
D
1
2
3
4
5
6

Flow
Time
After runPressure shock
Pressure shock suppression active as specified by the time entered
Pressure shock suppression inactive
Valve closes
Flow falls below the on-value of the low flow cut off: pressure shock suppression is activated
The time entered has elapsed: pressure shock suppression is deactivated
The actual flow value is now displayed and output
On-value for low flow cut off
Off-value for low flow cut off

Low flow cut off
Navigation

 Guidance → Commissioning

Low flow cut offNavigation

 Guidance → Commissioning → Low flow cut off

Description

Select process variable for low flow cut off to activate low flow cut off.

Selection• Off• Volume flow
• Mass flow

Additional information

Description

18

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn-value entered
Off-value entered

On value low flow cutoff

Navigation

 Guidance → Commissioning → On value

Description

Enter on value to switch on low flow cut off.
Value = 0: No low flow cut off
Value > 0: Low flow cut off is activatedUser entry

Positive floating-point number

Off value low flow cutoff

Navigation

 Guidance → Commissioning → Off value

Description

Enter off value to switch off low flow cut off. The off value is entered as a positive
hysteresis with respect to the on value.

User entry

0 to 100.0 %Endress+Hauser

19

"Guidance" menu

Proline Promag 10 HART

Empty pipe detection
Navigation

 Guidance → Commissioning

Empty pipe detection



Navigation

 Guidance → Commissioning → Empty pipe det.

Description

Switch empty pipe detection on or off. Switch on empty pipe detection to detect a partiallyfilled or empty measuring tube.

Selection

• Off
• On

Empty pipe adjust value

Navigation

 Guidance → Commissioning → Empty pipe value

Description

Displays adjustment value when the measuring tube is empty.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

Full pipe adjust value

Navigation

 Guidance → Commissioning → Full pipe value

Description

Displays adjustment value when the measuring tube is full.
NOTE
Users logged on in the Service role have write access!

User interface

20

Positive floating-point number

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

2.1.6

Current output

Navigation

 Guidance → Commissioning

Process variable current output
Navigation

 Guidance → Commissioning → Proc.var. outp

Description

Select process variable for current output

Selection• Off *• Mass flow
• Conductivity *• Corrected conductivity *• Volume flow
• Temperature *• Noise *• Coil current shot time *Current range output



Navigation

 Guidance → Commissioning → Curr.range out

Description

Select the current range for the measured value output and the upper and lower fault
condition signal level.
Additional information:- The measured value range is specified in the "Lower range value output " parameter andthe "Upper range value output " parameter.
- If the measured value lies outside the scaled measured value range, diagnostic message"441 Current output faulty" is generated.
- In the event of a device alarm, the current output adopts the behavior specified in the"Failure behavior current output " parameter.

Selection• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)• Fixed value

Additional information

Selection• 4...20 mA NE (3.8...20.5 mA) option
Select this option to set the current range in accordance with NAMUR recommendation
NE43.
• Fixed value option
Select this option to set the current output to a current value instead of a range.
The current value is defined in the Fixed current parameter (→  24).

*

Visibility depends on order options or device settings

Endress+Hauser

21

"Guidance" menu

Proline Promag 10 HART

The graphic shows the relationship between the current range for the output of theprocess value and the lower and upper alarm levels:
2

1

3

I [mA]
A0034351

1
2
3

Current range for process value
Lower level for signal on alarm
Upper level for signal on alarm

Selection (current range for process value) Lower level for signal on
alarm

Upper level for signal on
alarm

4...20 mA NE (3.8...20.5 mA)> 21.5 mA< 3.6 mA

4...20 mA US (3.9...20.8 mA)
4...20 mA (4... 20.5 mA)

Lower range value output

Navigation

 Guidance → Commissioning → Low.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter lower range value for the measured value range.
Additional information:- Depending on the setting selected for the "Measuring mode current output " parameter,
the value specified for this parameter and the "Upper range value output " parameter musthave the same algebraic sign or not.
- As a rule, the lower range value is scaled to be lower than the upper range value. As aresult, the behavior of the current output is proportional to the process variable assigned.
If the lower range value is scaled to be higher than the upper range value, then the
behavior of the current output will be inversely proportional to the process variableassigend.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter (→  97).
Example: Measuring mode with "Forward flow" option• Lower range value output parameter (→  22) = not equal to zero flow (e.g. –250
m3/h)
• Upper range value output parameter (→  23) = not equal to zero flow (e.g. +750
m3/h)
• Calculated current value = 8 mA at zero flow

22

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

I [mA]
20
84Q-250 0+750

11A0013757

Q
I
1

Flow
Current
Measuring range is exceeded or undershot

The linear operational range of the measuring device is defined by the values entered for
the Lower range value output parameter (→  22) and Upper range value output
parameter (→  23), and by the selected current range.
Example: Measuring mode with the "Forward/Reverse flow" option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

Q
I
1
2
3

Flow
Current
Value assigned to 0/4 mA current
Forward flow
Reverse flow

The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the Lower range value output parameter (→  22)
and Upper range value output parameter (→  23) must have the same algebraic sign.
The value for the Upper range value output parameter (→  23) (e.g. reverse flow)corresponds to the mirrored value for the Upper range value output parameter(→  23) (e.g. forward flow).
Example: Measuring mode with the "Reverse flow compensation" option
If flow is characterized by severe fluctuations (e.g. when using reciprocating pumps), flow
components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.

Upper range value output

Navigation

 Guidance → Commissioning → Upp.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Endress+Hauser

23

"Guidance" menu

Proline Promag 10 HART

Description

Enter upper range value for the measured value range.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter: Lower range value output parameter (→  22):

Damping current output

Navigation

 Guidance → Commissioning → Damp.curr.outp

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter time constant (PT1 element) to set the reaction time of the output signal tofluctuations in the measured value caused by process conditions.
Additional information:- The smaller the time constant the faster the output reacts to fluctuations in themeasured value.
- If the time constant is 0, damping is deactivated.

User entry

0.0 to 999.9 s

Fixed current

Navigation

 Guidance → Commissioning → Fixed current

Prerequisite

In the Current range output parameter in the Current output 1 submenu, the Fixedvalue option is selected.

Description

Enter the value for the "Fixed value" option.

User entry

3.59 to 21.5 mA

Failure behavior current output

Navigation

 Guidance → Commissioning → Fail.behav.out

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

24

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Description

Specify how the output should behave in the event of a device alarm.

Selection• Min.
• Max.
• Last valid value• Actual value• Fixed value

Additional information

Selection• Min. option
The current output outputs the lower fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Max. option
The current output outputs the upper fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Last valid value option
The current output outputs the last valid value measured before the device alarm
occurred.
• Actual value option
The current output outputs the flow value currently measured. The device alarm isignored.
• Fixed value option
The current output outputs the value specified.
Additional information:
The value is specified in the "Failure current " parameter.

Failure current

Navigation

 Guidance → Commissioning → Fail. current

Prerequisite

In the Failure behavior current output parameter in the Current output 1 submenu, theFixed value option is selected.

Description

Enter the value for the "Fixed value" option in the "Failure behavior current output "parameter.

User entry

3.59 to 21.5 mA

Endress+Hauser

25

"Guidance" menu

Proline Promag 10 HART

2.1.7

Pulse/frequency/switch output

Navigation

 Guidance → Commissioning

Operating modeNavigation

 Guidance → Commissioning → Operating mode

Description

Select the operating mode for the output.

Selection• Pulse• Frequency• Switch

Additional information

Selection• Pulse option
Quantitatively proportional pulse with pulse width to be configured. Whenever the pulse
value for the specified process variable has been reached, a pulse is emitted, the durationof which is set within the "Pulse width" parameter.
Additional information:
The process variable for the pulse output is specified in the "Assign pulse output "parameter.
• Frequency option
The output frequency is proportional to the value for the process variable assigned, witha pulse-to-interval ratio of 1:1.
Additional information:
The process variable for the frequency output is specified in the "Assign frequency
output" parameter.
• Switch option
Indicates when the state of the device changes, e.g. when a specified limit value isreached or an alarm or warning is triggered.
Additional information:- The switch output can be in one of two states: either it is conductive or it is nonconductive.
- When the function assigned to the switch output is triggered, the switch output will
depending on the output configuration either be continuously conductive or continuously
non-conductive.
- The switch output is used to display diagnostic information at the system level, e.g. byconnecting a lamp that lights up when the function assigned is triggered.
"Pulse" option
Example• Flow rate approx. 100 g/s• Pulse value 0.1 g• Pulse width 0.05 ms• Pulse rate 1 000 pulse/s

26

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

U[V]

B<P
B

24
t

0
P

A0026883

1
BP

Quantity-proportional pulse (pulse value) with pulse width to be configured
Pulse width entered
Pauses between the individual pulses"Frequency" option
Example• Flow rate Q approx. 100 g/s• Min. frequency (fmin) 0 Hz
• Max. frequency (fmax) 1000 Hz• Flow rate at min. frequency (Qmin) 0 g/s• Flow rate at max. frequency (Qmax) 1000 g/s• Output frequency (fout) approx. 100 Hz
fout = fmin + Q × [(fmax - fmin)/(Qmax - Qmin)] =0 Hz + 100 g/s × [(1000 Hz - 0 Hz)/(1000 g/s - 0 g/s )] = 100 Hz
U[V]

fout
T!T"

T!= T"

24t

0

A0026886

2

Flow-proportional frequency output"Switch" option
ExampleAlarm response without alarm
U[V]
24
0

t
A0026884

3

No alarm, high level

ExampleAlarm response in case of alarm

Endress+Hauser

27

"Guidance" menu

Proline Promag 10 HART

U[V]
24
0

t
A0026885

4

Alarm, low level

Frequency output
Navigation

 Guidance → Commissioning

Assign frequency output

Navigation

 Guidance → Commissioning → Assign freq.

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Select process variable for frequency output.

Selection• Off• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Noise *• Coil current shot time *Minimum frequency value

Navigation

 Guidance → Commissioning → Min. freq. value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter the minimum frequency for the frequency range.
Additional information:
The lower range value for the measured value range is specified in the "Measuring value atminimum frequency" parameter.

User entry*

28

0.0 to 10 000.0 Hz

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Measuring value at minimum frequency

Navigation

 Guidance → Commissioning → Val. at min.freq

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter lower range value for the measured value range.
Additional information:- Depending on the setting selected for the "Measuring mode" parameter, the value
specified for this parameter and the "Measuring value at maximum frequency" parametermust have the same algebraic sign or not.
- As a rule, the lower range value is scaled to be lower than the upper range value. As aresult, the behavior of the frequency output is proportional to the process variableassigned. If the lower range value is scaled to be higher than the upper range value, thenthe behavior of the frequency output will be inversely proportional to the process variableassigend.

User entry

Signed floating-point number

Maximum frequency value

Navigation

 Guidance → Commissioning → Max. freq. value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter the maximum frequency for the measured value output.
Additional information:
The upper range value for the measured value range that corresponds to the maximum
frequency is specified in the "Measuring value at maximum frequency" parameter.

User entry

0.0 to 10 000.0 Hz

Measuring value at maximum frequency

Navigation

 Guidance → Commissioning → Val. at max.freq

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter upper range value for the measured value range.

User entry

Signed floating-point number

Endress+Hauser

29

"Guidance" menu

Proline Promag 10 HART

Failure modeNavigation

 Guidance → Commissioning → Failure mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Specify how the output should behave in the event of a device alarm.
Additional information:
For safety reasons, it is recommended that the behavior of the output in the event of adevice alarm be predefined.

Selection• Actual value• Defined value• 0 Hz

Additional information

Selection• Actual value option
The frequency output outputs the flow value currently measured.
• Defined value option
The frequency output outputs the value specified.
Additional information:
The value is specified in the "Failure frequency" parameter.
• 0 Hz option
In the event of a device alarm, the frequency output outputs 0 Hz.

Failure frequency

Navigation

 Guidance → Commissioning → Failure freq.

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter the value for the "Defined value" option in the "Failure mode" parameter.

User entry

0.0 to 12 500.0 Hz

30

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Pulse output
Navigation

 Guidance → Commissioning

Assign pulse output 1



Navigation

 Guidance → Commissioning → Assign pulse 1

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Select process variable for pulse output.

Selection• Off• Volume flow
• Mass flow

Pulse widthNavigation

 Guidance → Commissioning → Pulse width

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Specify the duration of the output pulse.
Additional information:
The maximum pulse rate is defined by fmax = 1 / (2 × pulse width). The interval between
two pulses (P) is at least as long as the specified pulse width (B).
The maximum flow is defined by Qmax = fmax × pulse value. If the flow exceeds theselimit values, the measuring device displays the diagnostic message "443 Pulse output
faulty".
Example:- Pulse value: 0.1 g- Pulse width: 0.1 ms- fmax: 1 / (2 × 0.1 ms) = 5 kHz- Qmax: 5 kHz × 0.1 g = 0.5 kg/s

User entry

Endress+Hauser

0.05 to 2 000 ms

31

"Guidance" menu

Additional information

Proline Promag 10 HART

Description
U[V]

U[V]

B<P

B=P
B

B

t
P

t
P
A0026882

B
P

Pulse width entered
Pauses between the individual pulses

Value per pulseNavigation

 Guidance → Commissioning → Value per pulse

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Enter the measured value to which a pulse corresponds.
Additional information:
Weighting of the pulse output with a quantity.
The lower the pulse value, the– better the resolution.
– higher the frequency of the pulse response.

User entry

Signed floating-point number

Switch output
Navigation

 Guidance → Commissioning

Switch output function



Navigation

 Guidance → Commissioning → Switch out funct

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Assign a function to the switch output.
Additional information:- The state of the switch output (on or off) when the assigned function is triggered can beinverted in the "Invert output signal" parameter- The "Invert output signal" parameter is not available for all devices.

32

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Selection

• Off
• On
• Diagnostic behavior• Limit• Flow direction check• Status

Additional information

Selection• Off option
The switch output is permanently switched off (open, non-conductive).
• On option
The switch output is permanently switched on (closed, conductive).
• Diagnostic behavior option
The switch output is switched on (closed, conductive), if there is a pending diagnostic
event of the assigned behavioral category.
• Limit option
The switch output is switched on (closed, conductive), if a limit value specified for theprocess variable is reached.
• Flow direction check option
The switch output is switched on (closed, conductive), when the flow direction changes(forward or reverse flow).
• Status option
The switch output is switched on (closed, conductive) to indicate the device status for theselected detection method, e.g. empty pipe detection or low flow cut off.

Assign diagnostic behavior

Navigation

 Guidance → Commissioning → Assign diag. beh

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

The switch output is switched on (closed, conductive), if there is a pending diagnostic
event of the assigned behavioral category.

Selection• Alarm• Alarm or warning• Warning

Additional information

Selection• Alarm option
The switch output is only switched on for diagnostic events of the "Alarm" category.
• Alarm or warning option
The switch output is switched on for diagnostic events of the "Alarm" or "Warning"
category.
• Warning option
The switch output is only switched on for diagnostic events of the "Warning" category.

Endress+Hauser

33

"Guidance" menu

Proline Promag 10 HART

Assign limitNavigation

 Guidance → Commissioning → Assign limit

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Select the process variable to monitor in case the specified limit value is exceeded. If a limit
value for the selected process variable is exceeded, the output is switched on.

Selection• Off• Volume flow
• Mass flow
• Flow velocity• Conductivity *• Corrected conductivity *• Totalizer 1• Totalizer 2• Totalizer 3• Temperature *• Sensor electronics temperature (ISEM)

Additional information

Switch-on point > switch-off point
Behavior of the status output if switch-on point > switch-off point:• Process variable > switch-on point: transistor is conductive• Process variable < switch-off point: transistor is not conductive
A
1
2
t
B
3
4

t
A0026891

1
2
3
4
A
B

Switch-on point
Switch-off point
Conductive
Non-conductive
Process variable
Status output

Switch-on point < switch-off point
Behavior of the status output if switch-on point < switch-off point:• Process variable < switch-on point: transistor is conductive• Process variable > switch-off point: transistor is not conductive*

34

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

A
1
2
t
B
3
4

t
A0026892

1
2
3
4
A
B

Switch-on point
Switch-off point
Conductive
Non-conductive
Process variable
Status output

Switch-on point = switch-off point
Behavior of the status output if switch-on point = switch-off point:• Process variable > switch-on point: transistor is conductive• Process variable < switch-off point: transistor is not conductive
A
1
t
B
2
3

t
A0026893

1
2
3
A
B

Switch-on point = switch-off point
Conductive
Non-conductive
Process variable
Status output

Switch-on value

Navigation

 Guidance → Commissioning → Switch-on value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter limit value for the switch-on point (process variable > switch-on value = closed,
conductive).
Additional information:
To use a hysteresis: Switch-on point > Switch-off point.

User entry

Endress+Hauser

Signed floating-point number

35

"Guidance" menu

Proline Promag 10 HART

Switch-off value

Navigation

 Guidance → Commissioning → Switch-off value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter limit value for the switch-off point (process variable < switch-off value = open, nonconductive).
Additional information:
To use a hysteresis: Switch-on point > Switch-off point.

User entry

Signed floating-point number

Switch-on delay

Navigation

 Guidance → Commissioning → Switch-on delay

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter delay before the switch output is switched on.

User entry

0.0 to 100.0 s

Switch-off delay

Navigation

 Guidance → Commissioning → Switch-off delay

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter delay before the switch output is switched off.

User entry

0.0 to 100.0 s

Assign status



Navigation

 Guidance → Commissioning → Assign status

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

36

Endress+Hauser

Proline Promag 10 HART

Description"Guidance" menu

Select the device status to display for the switch output.
Additional information:
If the switch on point for the selected detection method is reached, the output isconductive. Otherwise, the switch output is non-conductive.

Selection• Empty pipe detection• Low flow cut off

Failure modeNavigation

 Guidance → Commissioning → Failure mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Specify how the output should behave in the event of a device alarm.
Additional information:
For safety reasons, it is recommended that the behavior of the output in the event of adevice alarm be predefined.

Selection• Actual status• Open• Closed

Additional information

Selection• Actual status option
In the event of a device alarm, the issue is ignored and the switch output adopts the
behavior currently specified for the "Switch output function" parameter.
• Open option
In the event of a device alarm, the switch output's transistor is set to "non-conductive".

2.1.8
Navigation

Display Guidance → Commissioning

Value 1 display

Navigation

 Guidance → Commissioning → Value 1 display

Description

Select the measured value that is displayed first on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Endress+Hauser

37

"Guidance" menu

Selection

Proline Promag 10 HART

• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 2 display

Navigation

 Guidance → Commissioning → Value 2 display

Description

Select the measured value that is shown second on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 3 display

Navigation

 Guidance → Commissioning → Value 3 display

Description

Select the measured value that is shown third on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection*

38• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Value 4 display

Navigation

 Guidance → Commissioning → Value 4 display

Description

Select the measured value that is shown fourth on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Display damping

Navigation

 Guidance → Commissioning → Display damping

Description

Enter time constant (PT1 element) to set reaction time of the display to fluctuations in themeasured value.
Additional information:- The smaller the time constant the faster the display reacts to fluctuations in themeasured value.
- If the time constant is set to 0, damping is deactivated.

User entry

0.0 to 999.9 s

2.1.9Date/time

Navigation

 Guidance → Commissioning

Time format



Navigation

 Guidance → Commissioning → Time format

Description

Select time format.

*

Visibility depends on order options or device settings

Endress+Hauser

39

"Guidance" menu

Selection

Proline Promag 10 HART

• 24 h• 12 h AM/PMTime zoneNavigation

 Guidance → Commissioning → Time zone

Description

Select the time zone. Every time the time zone is changed, a logbook entry is created.

Selection

40

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+14:00

Endress+Hauser

Proline Promag 10 HART

"Guidance" menu

Set date/timeNavigationDescription

Set the date and local time. Every time the date or time is changed, a logbook entry iscreated.

User entry

Date and time

Endress+Hauser

Guidance → Commissioning → Set date/time

41

"Diagnostics" menu

Proline Promag 10 HART

3

"Diagnostics" menu

Troubleshooting and preventive maintenance – settings for device behavior during process
and device events as well as assistance and measures for diagnostic purposes.
Navigation

 Diagnostics

Diagnostics

42‣ Active diagnostics→  43‣ Diagnostic list→  45‣ Simulation→  48‣ Heartbeat Technology→  53‣ Diagnostic settings→  54Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

3.1

"Actual diagnostics" submenu

Navigation

 Diagnostics → Active diagnos.

‣ Active diagnostics
Actual diagnostics→  43

Timestamp→  43

Previous diagnostics→  43

Timestamp→  44

Operating time from restart→  44

Operating time→  44

Actual diagnostics
Navigation

 Diagnostics → Active diagnos. → Actual diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the currently active diagnostic message.
If there is more than one pending diagnostic event, the message for the diagnostic eventwith the highest priority is displayed.

User interface

Positive integer

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp for the currently active diagnostic message.

User interface

Days (d), hours (h), minutes (m), seconds (s)Previous diagnostics
Navigation

 Diagnostics → Active diagnos. → Prev.diagnostics

Prerequisite

At least two diagnostic events have already occurred.

Endress+Hauser

43

"Diagnostics" menu

Proline Promag 10 HART

Description

Displays the diagnostic message for the last diagnostic event that has ended.

User interface

Positive integer

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp of the diagnostic message generated for the last diagnostic eventthat has ended.

User interface

Days (d), hours (h), minutes (m), seconds (s)Operating time from restart
Navigation

 Diagnostics → Active diagnos. → Time fr. restart

Description

Indicates how long the device has been in operation since the last time the device wasrestarted.

User interface

Days (d), hours (h), minutes (m), seconds (s)Operating time
Navigation

 Diagnostics → Active diagnos. → Operating time

Description

Indicates how long the device has been in operation.

User interface

Days (d), hours (h), minutes (m), seconds (s)44Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

3.2

"Diagnostic list" submenu

Navigation

Diagnostics → Diagnostic list

‣ Diagnostic list
Diagnostics 1→  45

Timestamp→  45

Diagnostics 2→  46

Timestamp→  46

Diagnostics 3→  46

Timestamp→  46

Diagnostics 4→  46

Timestamp→  47

Diagnostics 5→  47

Timestamp→  47

Diagnostics 1
Navigation

Diagnostics → Diagnostic list → Diagnostics 1

Description

Displays the currently active diagnostic message with the highest priority.

User interface

Positive integer

Timestamp
NavigationDescription

Displays the timestamp for the diagnostic message with the highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Diagnostics → Diagnostic list → Timestamp

45

"Diagnostics" menu

Proline Promag 10 HART

Diagnostics 2
Navigation

Diagnostics → Diagnostic list → Diagnostics 2

Description

Displays the currently active diagnostic message with the second highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the second highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 3
Navigation

Diagnostics → Diagnostic list → Diagnostics 3

Description

Displays the currently active diagnostic message with the third highest priority.

User interface

Positive integer

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the third highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 4
NavigationDescription

Displays the currently active diagnostic message with the fourth highest priority.

User interface

Positive integer

46

Diagnostics → Diagnostic list → Diagnostics 4

Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

Timestamp
Navigation

Diagnostics → Diagnostic list → Timestamp

Description

Displays the timestamp for the diagnostic message with the fourth highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Diagnostics 5
Navigation

Diagnostics → Diagnostic list → Diagnostics 5

Description

Displays the currently active diagnostic message with the fifth-highest priority.

User interface

Positive integer

Timestamp
NavigationDescription

Displays the timestamp for the diagnostic message with the fifth highest priority.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Diagnostics → Diagnostic list → Timestamp47

"Diagnostics" menu

Proline Promag 10 HART

3.3

"Simulation" submenu

Navigation

 Diagnostics → Simulation‣ Simulation
Assign simulation process variable→  48Process variable value→  49Current output 1 simulation→  49Current output value→  49Frequency output 1 simulation→  50Frequency output 1 value→  50

Pulse output simulation 1→  50

Pulse value 1→  51

Switch output simulation 1→  51

Switch state 1→  51

Device alarm simulation→  52

Diagnostic event category→  52

Diagnostic event simulation→  52

Assign simulation process variable
Navigation

 Diagnostics → Simulation → Assign proc.var.

Description

Select a process variable for the simulation, thereby activating it.

Selection• Off• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *

*

48



Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

Additional information"Diagnostics" menu

Description
The display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

Process variable value

Navigation

 Diagnostics → Simulation → Proc. var. value

Description

Enter the simulation value for the selected process variable. Processing of measured values
downstream as well as the signal ouput follow this value. In this way, it is possibe to verify
whether the measuring device has been configured correctly.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User entry

Signed floating-point number

Current output 1 simulation



Navigation

 Diagnostics → Simulation → Curr.outp 1 sim.

Description

Switch simulation of the current output on or off.
Additional information:
When simulation is on, the current output signal is set to the value specified in the"Current output value" parameter.

Selection

• Off
• On

Additional information

Description
The display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

Current output value

Navigation

 Diagnostics → Simulation → Curr.outp val.

Description

Enter the current value for the simulation. In this manner, it is possible to verify thecurrent output is configured correctly and downstream processing units are functioning
properly.
Additional information:
The valid input range is determined by the "Current range output " parameter.

User entry

Endress+Hauser

3.59 to 21.5 mA

49

"Diagnostics" menu

Proline Promag 10 HART

Frequency output 1 simulation
Navigation

 Diagnostics → Simulation → Freq.outp 1 sim.

Description

Switch simulation of the frequency output on or off.

Selection

• Off
• On

Additional information

DescriptionThe display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

Frequency output 1 value

Navigation

 Diagnostics → Simulation → Freq.outp 1 val.

Description

Enter the frequency value for the simulation. In this manner, it is possible to verify thefrequency output is configured correctly and downstream processing units are functioning
properly.

User entry

0.0 to 12 500.0 Hz

Pulse output simulation 1

Navigation

 Diagnostics → Simulation → Puls.outp.sim. 1

Description

Switch simulation of the pulse output on or off.

Selection• Off• Fixed value• Down-counting value

Additional information

Selection• Off option
Simulation of the pulse output is switched off. The device is in standard operation modeor another process variable is being simulated.
• Fixed value option
Pulses are emitted continuously with the pulse width specified in the "Pulse width"parameter.
• Down-counting value option
The number of pulses specified in the "Pulse value " parameter are emitted.
Description
The display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

50

Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

Pulse value 1Navigation

 Diagnostics → Simulation → Pulse value 1

Description

Enter the number of pulses to simulate the pulse output. In this manner, it is possible toverify the pulse output is configured correctly and downstream processing units arefunctioning properly.

User entry

0 to 65 535

Switch output simulation 1

Navigation

 Diagnostics → Simulation → Switch sim. 1

Description

Switch simulation of the switch output on or off.
Additional information:
When simulation is on, the switch output is set to the state specified in the "Switch state "parameter.

Selection

• Off
• On

Additional information

Description
The display alternates between the measured value and a diagnostics message of the"function check" category (C) when simulation is active.

Switch state 1Navigation

 Diagnostics → Simulation → Switch state 1

Description

Select a switch state. In this manner, it is possible to verify the switch output is configured
correctly and downstream processing units are functioning properly.

Selection• Open• Closed

Additional information

Selection• Open option
The switch output is not conductive.
• Closed option
The switch output is conductive.

Endress+Hauser

51

"Diagnostics" menu

Proline Promag 10 HART

Device alarm simulation



Navigation

 Diagnostics → Simulation → Dev. alarm sim.

Description

Switch the device alarm simulation on or off.
While simulation is in progress, the display alternates between the measured value and adiagnostic message of the Function Check (C) category.

Selection

• Off
• On

Diagnostic event category



Navigation

Diagnostics → Simulation → Event category

Description

Select the category of diagnostic events that should be available for selection in the"Diagnostic event simulation" parameter.

Selection• Sensor• Electronics• Configuration• ProcessDiagnostic event simulation



Navigation

 Diagnostics → Simulation → Diag. event sim.

Description

Select the diagnostic event to simulate.

Selection

Off

52

Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

3.4

"Heartbeat Technology" submenu

The Heartbeat Technology submenu (→  53) is only available with the optional"Heartbeat Verification + Monitoring" application package.
• Order code for: Application package• Option: EB "Heartbeat Verification + Monitoring"
Detailed information and all descriptions of the device parameters of the application
package are available in the "Heartbeat Verification + Monitoring" Special
Documentation
Navigation

 Diagnostics → HBT‣ Heartbeat Technology‣ Performing verification‣ Verification results‣ Heartbeat Monitoring

Endress+Hauser

53

"Diagnostics" menu

Proline Promag 10 HART

3.5

"Diagnostic settings" submenu

Navigation

 Diagnostics → Diag. settings‣ Diagnostic settings

3.5.1‣ Properties→  54‣ Diagnostic configuration→  54"Properties" submenu

Navigation

 Diagnostics → Diag. settings → Properties‣ Properties
Alarm delay→  54

Alarm delay

Navigation

 Diagnostics → Diag. settings → Properties → Alarm delay

Description

Enter a duration for the alarm delay. When a diagnostic event of the "Alarm" category
occurs, the diagnostic message is not generated until the delay has elapsed.

User entry

0 to 60 s

3.5.2
Navigation"Diagnostic configuration" submenu
 Diagnostics → Diag. settings → Diag. config.

‣ Diagnostic configuration

54‣ Sensor→  55‣ Electronics→  56‣ Configuration→  59‣ Process→  62Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

"Sensor" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Sensor‣ Sensor
Assign behavior of diagnostic no. 043→  55

Event category 043→  55Assign behavior of diagnostic no. 043
Navigation

 Diagnostics → Diag. settings → Diag. config. → Sensor → Diagnostic no. 043

Description

Select behavior for diagnostic event "043 Sensor short circuit detected".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 043



Navigation

 Diagnostics → Diag. settings → Diag. config. → Sensor → Event category 043

Description

Select event category (status signal) for diagnostic event "043 Sensor short circuit
detected".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Endress+Hauser

55

"Diagnostics" menu

Proline Promag 10 HART

"Electronics" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics‣ Electronics
Assign behavior of diagnostic no. 230→  56

Event category 230→  56

Assign behavior of diagnostic no. 231→  57

Event category 231→  57

Assign behavior of diagnostic no. 302→  57

Event category 302→  57

Assign behavior of diagnostic no. 376→  58

Event category 376→  58Assign behavior of diagnostic no. 377→  59Event category 377→  59Assign behavior of diagnostic no. 230

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 230

Description

Select behavior for diagnostic event "230 Date/time incorrect".

Selection• Alarm• Warning• Logbook entry only

Event category 230

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Event category 230

Description

Select event category (status signal) for diagnostic event "230 Date/time incorrect".

56

Endress+Hauser

Proline Promag 10 HART

Selection"Diagnostics" menu

• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 231
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 231

Description

Select behavior for diagnostic event "231 Date/time not available".

Selection• Alarm• Warning• Logbook entry only

Event category 231

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Event category 231

Description

Select event category (status signal) for diagnostic event "231 Date/time not available".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 302
Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 302

Description

Select behavior for diagnostic event "302 Device verification active".

Selection• Off• Warning• Logbook entry only

Event category 302



Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Event category 302

Description

Select event category (status signal) for diagnostic event "302 Device verification active".

Endress+Hauser



57

"Diagnostics" menu

Selection

Proline Promag 10 HART

• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 376

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 376

Description

Select behavior for diagnostic event "376 Main electronics faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 376

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Event category 376

Description

Select event category (status signal) for diagnostic event "376 Sensor electronics (ISEM)
faulty".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)58

Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

Assign behavior of diagnostic no. 377

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Diagnostic no. 377

Description

Select behavior for diagnostic event "377 Sensor electronics (ISEM) faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 377

Navigation

 Diagnostics → Diag. settings → Diag. config. → Electronics → Event category 377

Description

Select event category (status signal) for diagnostic event "377 Sensor electronics (ISEM)
faulty".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)"Configuration" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration‣ Configuration

Endress+Hauser

Assign behavior of diagnostic no. 441→  60

Event category 441→  6059

"Diagnostics" menu

Proline Promag 10 HART

Assign behavior of diagnostic no. 442→  61

Event category 442→  61

Assign behavior of diagnostic no. 443→  61

Event category 443→  62

Assign behavior of diagnostic no. 441

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Diagnostic no. 441

Description

Select behavior for diagnostic event "441 Current output faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 441

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Event category 441

Description

Select event category (status signal) for diagnostic event "441 Current output faulty".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)60

Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

Assign behavior of diagnostic no. 442

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Diagnostic no. 442

Description

Select behavior for diagnostic event "442 Frequency output faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 442

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Event category 442

Description

Select event category (status signal) for diagnostic event "442 Frequency output faulty".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 443

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Diagnostic no. 443

Description

Select behavior for diagnostic event "443 Pulse output faulty".

Selection• Off• Alarm• Warning• Logbook entry only

Endress+Hauser

61

"Diagnostics" menu

Additional information

Proline Promag 10 HART

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 443

Navigation

 Diagnostics → Diag. settings → Diag. config. → Configuration → Event category 443

Description

Select event category (status signal) for diagnostic event "443 Pulse output faulty".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)"Process" submenu
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process‣ Process62

Assign behavior of diagnostic no. 832→  63

Event category 832→  64

Assign behavior of diagnostic no. 833→  64

Event category 833→  64

Assign behavior of diagnostic no. 834→  65

Event category 834→  65Assign behavior of diagnostic no. 835→  66Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

Event category 835→  66

Assign behavior of diagnostic no. 842→  66

Event category 842→  67

Assign behavior of diagnostic no. 937→  67

Event category 937→  68Assign behavior of diagnostic no. 938→  68

Event category 938→  68Assign behavior of diagnostic no. 961→  69Event category 961→  69Assign behavior of diagnostic no. 962→  69Event category 962→  70Assign behavior of diagnostic no. 832
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 832

Description

Select behavior for diagnostic event "832 Electronics temperature too high".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Endress+Hauser

63

"Diagnostics" menu

Proline Promag 10 HART

Event category 832

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 832

Description

Select event category (status signal) for diagnostic event "832 Electronics temperature toohigh".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 833

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 833

Description

Select behavior for diagnostic event "833 Electronics temperature too low".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 833

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 833

Description

Select event category (status signal) for diagnostic event "833 Electronics temperature too
low".

64

Endress+Hauser

Proline Promag 10 HART

Selection"Diagnostics" menu

• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 834
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 834

Description

Select event category (status signal) for diagnostic event "834 Process temperature toohigh".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 834

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 834

Description

Select event category (status signal) for diagnostic event "834 Process temperature toohigh".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Endress+Hauser

65

"Diagnostics" menu

Proline Promag 10 HART

Assign behavior of diagnostic no. 835

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 835

Description

Select behavior for diagnostic event "835 Process temperature too low".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 835

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 835

Description

Select event category (status signal) for diagnostic event "835 Process temperature too
low".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 842

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 842

Description

Select behavior for diagnostic event "842 Process value above limit".

Selection• Off• Alarm• Warning• Logbook entry only

66

Endress+Hauser

Proline Promag 10 HART

Additional information"Diagnostics" menu

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 842

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 842

Description

Select event category (status signal) for diagnostic event "842 Process value above limit".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 937
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 937

Description

Select behavior for diagnostic event "937 Sensor symmetry".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.
Endress+Hauser

67

"Diagnostics" menu

Proline Promag 10 HART

Event category 937

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 937

Description

Select event category (status signal) for diagnostic event "937 Sensor symmetry".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 938

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 938

Description

Select behavior for diagnostic event "938 EMC interference".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 938

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 938

Description

Select event category (status signal) for diagnostic event "938 EMC interference".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)68

Endress+Hauser

Proline Promag 10 HART

"Diagnostics" menu

Assign behavior of diagnostic no. 961
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 961

Description

Select behavior for diagnostic event "961 Electrode potential out of specification".

Selection• Off• Alarm• Warning• Logbook entry only

Additional information

Selection

• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 961

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 961

Description

Select event category (status signal) for diagnostic event "961 Electrode potential out ofspecification".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Assign behavior of diagnostic no. 962
Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Diagnostic no. 962

Description

Select behavior for diagnostic event "962 Pipe empty".

Selection• Off• Alarm• Warning• Logbook entry only

Endress+Hauser

69

"Diagnostics" menu

Additional information

Proline Promag 10 HART

Selection• Off option
The diagnostic event is ignored and no diagnostic message is generated or logged.
• Alarm option
The device stops measuring. The signal outputs and totalizers assume the specified
alarm condition. A diagnostic message is generated.
• Warning option
The device continues measuring. The signal outputs and totalizers are not affected. A
diagnostic message is generated.
• Logbook entry only option
The device continues measuring. The diagnostic message is only displayed in the "Eventlogbook" submenu and does not alternate with the standard operational information
displayed.

Event category 962

Navigation

 Diagnostics → Diag. settings → Diag. config. → Process → Event category 962

Description

Select event category (status signal) for diagnostic event "962 Pipe empty".

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)70

Endress+Hauser

Proline Promag 10 HART

"Application" menu

4"Application" menu

Targeted optimization to the application – comprehensive device settings from sensor
technology to system integration for optimum application adaptation.
Navigation

 Application

Application

4.1‣ Measured values→  71‣ System units→  75‣ Totalizers→  80‣ Sensor→  85‣ Current output 1→  96‣ Pulse/frequency/switch output 1→  107‣ Communication→  125"Measured values" submenu

Navigation

 Application → Measured values‣ Measured values

Endress+Hauser

Mass flow→  72

Volume flow

→  72

Conductivity→  72

Corrected conductivity→  73Temperature→  73‣ Totalizer→  7371

"Application" menu

Proline Promag 10 HART

Mass flow
Navigation

 Application → Measured values → Mass flow

Description

Displays the mass flow currently calculated.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Signed floating-point number

Volume flow
Navigation

 Application → Measured values → Volume flow

Description

Displays the volume flow currently measured.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Signed floating-point number

Conductivity
Navigation

 Application → Measured values → Conductivity

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the conductivity currently measured.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

72

Positive floating-point number

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Corrected conductivity
Navigation

 Application → Measured values → CorrConductivity

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays the currently measured conductivity compensated for temperature.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Positive floating-point number

Temperature
Navigation

 Application → Measured values → Temperature

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Displays the medium temperature currently measured.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User interface

Positive floating-point number

4.1.1"Totalizer" submenu

Navigation

 Application → Measured values → Totalizer‣ Totalizer

Endress+Hauser

Totalizer value 1 to n

→  74

Totalizer overflow 1 to n

→  74

73

"Application" menu

Proline Promag 10 HART

Totalizer value 1 to n

Navigation

 Application → Measured values → Totalizer → Totalizer val. 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Displays the current totalizer counter.
Additional information:
Since the operating tool cannot display figures with more than 7 digits, the current counter
above this range equals the sum of the totalizer counter plus the overflow displayed for the"Totalizer overflow " parameter.
Example for how to calculate the current totalizer counter when the value exceeds the 7digit display range limit of the operating tool:- Value of "Totalizer value " parameter: 1,968,457 m³- Value of "Totalizer overflow " parameter: 1 × 107 m³ = 10,000,000 m³- Current totalizer reading: 11,968,457 m³
In the event of an error, the totalizer behaves as specified in the "Failure mode" parameter.

User interface

Signed floating-point number

Totalizer overflow 1 to n

Navigation

 Application → Measured values → Totalizer → Tot. overflow 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Displays the current totalizer overflow.
Additional information:
If the current totalizer counter exceeds the operating tool's maximum numerical display
range of 7 digits, the amount above this range is expressed as an overflow. The current
totalizer counter therefore equals the sum of the overflow and the totalizer value displayedin the "Totalizer value " parameter.
Example of how to calculate the current totalizer counter when the value exceeds the 7digit display limit of the operating tool:- Value of "Totalizer value " parameter: 1,968,457 m³- Value of "Totalizer overflow " parameter: 1 × 10^7 m³ = 10,000,000 m³- Current totalizer reading: 11,968,457 m³

User interface

74–32 000.0 to 32 000.0

Endress+Hauser

Proline Promag 10 HART

"Application" menu

4.2

"System units" submenu

Navigation

 Application → System units‣ System units
Volume flow unit

→  75

Volume unit

→  76Mass flow unit

→  77Mass unit→  77

Density unit

→  78Temperature unit

→  78

Conductivity unit

→  78Volume flow unitNavigation

 Application → System units → Volume flow unit

Description

Select volume flow unit.

Endress+Hauser

75

"Application" menu

Selection

Proline Promag 10 HART

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dUS units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• MMft³/s• MMft³/min• MMft³/h• Mft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)Volume unitNavigation

 Application → System units → Volume unit

Description

Select volume unit.

76

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)Endress+Hauser

Proline Promag 10 HART

Selection"Application" menu

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega

US units• af• ft³• Mft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;beer)• bbl (us;tank)

Mass flow unitNavigation

 Application → System units → Mass flow unit

Description

Select mass flow unit.

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

Mass unitNavigation

 Application → System units → Mass unit

Description

Select mass unit.

Selection

SI units
•g
• kg
•t

Endress+Hauser

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

US units• oz• lb• STon77

"Application" menu

Proline Promag 10 HART

Density unitNavigation

 Application → System units → Density unit

Description

Select density unit.

Selection

SI units• g/cm³• g/m³• kg/l• kg/dm³• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

US units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)Temperature unitNavigation

 Application → System units → Temperature unit

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Select temperature unit.

Selection

SI units• °C•K

US units• °F• °R

Conductivity unitNavigation

 Application → System units → Conductiv. unit

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

78

Select conductivity unit.

Endress+Hauser

Proline Promag 10 HART

Selection

Endress+Hauser"Application" menu

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/m
• kS/m• MS/m79

"Application" menu

Proline Promag 10 HART

4.3

"Totalizers" submenu

Navigation

 Application → Totalizers‣ Totalizers

4.3.1‣ Totalizer handling→  80‣ Totalizer 1 to n

→  80"Totalizer handling" submenu

Navigation

 Application → Totalizers → Totalizer‣ Totalizer handlingReset all totalizers→  80Reset all totalizers
Navigation

 Application → Totalizers → Totalizer → Reset all tot.

Description

Reset all totalizers to "0" and restart the totaling process. All flow quantities thus far
totalized are thereby deleted.

Selection• Cancel• Reset + totalize

4.3.2"Totalizer 1 to n" submenu

Navigation

 Application → Totalizers → Totalizer 1 to n

‣ Totalizer 1 to n

80

Assign process variable→  81

Unit totalizer 1 to n

→  81

Totalizer operation mode→  82Control Totalizer 1 to n

→  82Endress+Hauser

Proline Promag 10 HART

"Application" menu

Preset value 1 to n

→  83

Failure mode→  83

Assign process variable

Navigation

 Application → Totalizers → Totalizer 1 to n → Assign variable

Description

Select process variable for totalizer.
Additional information:
If the option selected is changed, the device resets the totalizer to "0".

Selection• Off• Volume flow
• Mass flow

Unit totalizer 1 to n

Navigation

 Application → Totalizers → Totalizer 1 to n → Unit totalizer 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select process variable totalizer unit.

Selection

SI units• g*• kg *• t**

US units• oz *• lb *• STon *Visibility depends on order options or device settingsorSI units• cm³ *
• dm³ *
• m³ *• ml *• l*• hl *• Ml Mega **Endress+Hauser

US units• af *• ft³ *• Mft³ *• fl oz (us) *• gal (us) *• kgal (us) *• Mgal (us) *• bbl (us;liq.) *• bbl (us;beer) *• bbl (us;oil) *• bbl (us;tank) *Imperial units• gal (imp) *• Mgal (imp) *• bbl (imp;beer) *• bbl (imp;oil) *Visibility depends on order options or device settings

81

"Application" menu

Proline Promag 10 HART

or
Other units
None **

Additional information

Visibility depends on order options or device settings

Description
The unit is selected separately for each totalizer. The unit is independent of the option
selected in the System units submenu (→  75).
Selection
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  14).

Totalizer operation modeNavigation

 Application → Totalizers → Totalizer 1 to n → Operation mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Select totalizer calculation mode.

Selection• Net flow total• Forward flow total• Reverse flow total

Additional information

Selection• Net flow total option
The flow values in the forward and reverse flow directions are totalized and netted
against each other. Net flow is recorded in the flow direction.
• Forward flow total option
Only the flow in the forward flow direction is totalized.
• Reverse flow total option
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Control Totalizer 1 to n
Navigation

 Application → Totalizers → Totalizer 1 to n → Control Tot. 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Operate the totalizer.

82

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Selection• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• HoldAdditional information

Selection• Totalize option
The totalizer is started or continues running.
• Reset + hold option
The totaling process is stopped and the totalizer is reset to "0".
• Preset + hold option
The totaling process is stopped and the totalizer is set to the start value specified in the"Preset value " parameter.
• Reset + totalize option
The totalizer is reset to "0" and the totaling process is restarted.
• Hold option
Totalizing is stopped.

Preset value 1 to n
Navigation

 Application → Totalizers → Totalizer 1 to n → Preset value 1 to n

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify start value for totalizer.

User entry

Signed floating-point number

Additional information

Description
The unit of the selected process variable is specified for the totalizer in the Unit totalizer
parameter (→  14).
ExampleThis configuration is suitable for applications such as iterative filling processes with a fixed
batch quantity.

Failure modeNavigation

 Application → Totalizers → Totalizer 1 to n → Failure mode

Prerequisite

A process variable has been selected in the Assign process variable parameter in the
Totalizer 1 to n submenu.

Description

Specify how the totalizer should behave in the event of a device alarm.
Additional information:
The failsafe mode that applies to any other totalizers or outputs is specified separately inother parameters and is not impacted by this setting.

Endress+Hauser

83

"Application" menu

Proline Promag 10 HART

Selection• Stop• Actual value• Last valid value

Additional information

Selection• Stop option
The totalizer is stopped in the event of a device alarm.
• Actual value option
The totalizer continues to totalize based on the current value measured; the device alarmis ignored.
• Last valid value option
The totalizer continues to totalize based on the last valid value measured before thedevice alarm occurred.

84

Endress+Hauser

Proline Promag 10 HART

"Application" menu

4.4
Navigation"Sensor" submenu
 Application → Sensor‣ Sensor

4.4.1
Navigation‣ Process parameters→  85‣ Low flow cut off→  88‣ Empty pipe detection→  89‣ Sensor adjustment→  91‣ Calibration→  92‣ Electrode cleaning cycle→  94"Process parameters" submenu
 Application → Sensor → Process param.

‣ Process parameters

Endress+Hauser

Flow damping→  86

Flow override→  86

Conductivity measurement→  86

Conductivity temperature coefficient→  87

Conductivity damping time→  87

Temperature damping time→  88Fixed density→  8885

"Application" menu

Proline Promag 10 HART

Flow damping

Navigation

 Application → Sensor → Process param. → Flow damping

Description

Enter value for damping of the flow measured value in order to reduce the variability of
the flow measured value when exposed to interference.
Additional information:
The depth of the flow filter is determined by this setting. As the filter depth increases, sodoes the reaction time of the device.
- Value = 0: No damping. Damping of 0 is not recommended, as the measuring signal isthen so noisy that it is almost impossible to perform a measurement.
- Value > 0: Damping increases
Optimal damping depends on the measuring period.
Damping impacts the following measuring device variables:- Outputs- Low flow cut off- Totalizers

User entry

0 to 15

Flow override

Navigation

 Application → Sensor → Process param. → Flow override

Description

Stops the measuring process. Can be used for example when cleaning the pipeline.

Selection

• Off
• On

Additional information

Selection"On" option
Activates flow override. The diagnostic message "453 Flow override active" is generated.
Additional information:
Output values:- Temperature: Measurement continues- Totalizers 1 to 3: No longer totalize

Conductivity measurement

Navigation

 Application → Sensor → Process param. → Conduct. measur.

Prerequisite

Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option", option
CX (conductivity measurement)

86

Endress+Hauser

Proline Promag 10 HART

Description"Application" menu

Switch conductivity measurement on or off.
Additional information:
To be able to measure conductivity, the medium must have a minimum conductivity of 5
μS/cm.

Selection

• Off
• On

Conductivity temperature coefficientNavigation

 Application → Sensor → Process param. → Cond. temp.coeff

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Enter the temperature coefficient for conductivity.

User entry

Signed floating-point number

Conductivity damping timeNavigation

Application → Sensor → Process param. → ConductDampTime

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Enter time constant for conductivity damping (PT1 element):- Value = 0: No damping- Value > 0: Damping increases
Additional information:
Damping is implemented by means of a proportional transmission behavior with firstorder delay (PT1 element).

User entry

Endress+Hauser

0 to 999.9 s

87"Application" menu

Proline Promag 10 HART

Temperature damping timeNavigation

Application → Sensor → Process param. → TempDampingTime

Prerequisite

Temperature measurement is only optionally available for Promag H 10 (5HBB): Under
order code for "Functionality", option D (extended transmitter) and order code for "Sensor
option", option CI (medium temperature measurement)

Description

Enter time constant for damping the temperature value.

User entry

0 to 999.9 s

Fixed density

Navigation

 Application → Sensor → Process param. → Fixed density

Description

Enter a fixed value for the density.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

User entry

Positive floating-point number

4.4.2"Low flow cut off" submenu

Navigation

 Application → Sensor → Low flow cut off‣ Low flow cut off
Low flow cut off→  88On value low flow cutoff→  89Off value low flow cutoff→  89Low flow cut offNavigation

 Application → Sensor → Low flow cut off → Low flow cut off

Description

Select process variable for low flow cut off to activate low flow cut off.

Selection• Off• Volume flow
• Mass flow

88

Endress+Hauser

Proline Promag 10 HART

Additional information"Application" menu

Description
Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn-value entered
Off-value entered

On value low flow cutoff

Navigation

 Application → Sensor → Low flow cut off → On value

Description

Enter on value to switch on low flow cut off.
Value = 0: No low flow cut off
Value > 0: Low flow cut off is activatedUser entry

Positive floating-point number

Off value low flow cutoff

Navigation

 Application → Sensor → Low flow cut off → Off value

Description

Enter off value to switch off low flow cut off. The off value is entered as a positive
hysteresis with respect to the on value.

User entry

0 to 100.0 %4.4.3
Navigation"Partially filled pipe detection" submenu
 Application → Sensor → Partial pipe det‣ Empty pipe detection
Empty pipe detection

Endress+Hauser→  9089

"Application" menu

Proline Promag 10 HART

New adjustment→  90Empty pipe adjust value→  90

Full pipe adjust value→  91

Measured value EPD→  91

Empty pipe detection



Navigation

 Application → Sensor → Empty pipe det. → Empty pipe det.

Description

Switch empty pipe detection on or off. Switch on empty pipe detection to detect a partiallyfilled or empty measuring tube.

Selection

• Off
• On

New adjustment

Navigation

 Application → Sensor → Empty pipe det. → New adjustment

Description

Select empty pipe or full pipe adjustment to perform a new adjustment. To adjust empty
pipe detection, perform the empty pipe adjustment first and then the full pipe adjustment.
Additional information:
The measuring device is pre-adjusted at production using water (approx. 300 μS/cm). For
liquids that deviate from this conductivity, a new empty pipe and full pipe adjustment mustbe performed on site.

Selection• Cancel• Empty pipe adjust• Full pipe adjust

Empty pipe adjust value

Navigation

 Application → Sensor → Empty pipe det. → Empty pipe value

Description

Displays adjustment value when the measuring tube is empty.
NOTE
Users logged on in the Service role have write access!

User interface

90

Positive floating-point number

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Full pipe adjust value

Navigation

 Application → Sensor → Empty pipe det. → Full pipe value

Description

Displays adjustment value when the measuring tube is full.
NOTE
Users logged on in the Service role have write access!

User interface

Positive floating-point number

Measured value EPD
Navigation

 Application → Sensor → Empty pipe det. → Meas. value EPD

Description

Displays the value currently measured for empty pipe detection.

User interface

Positive floating-point number

4.4.4"Sensor adjustment" submenu

Navigation

 Application → Sensor → Sensor adjustm.

‣ Sensor adjustment
Installation direction→  91

Integration time→  92

Measuring period→  92

Installation direction



Navigation

 Application → Sensor → Sensor adjustm. → Install. direct.

Description

Select sign of flow direction

Selection• Forward flow
• Reverse flow

Endress+Hauser

91

"Application" menu

Proline Promag 10 HART

Integration timeNavigation

 Application → Sensor → Sensor adjustm. → Integration time

Description

Displays the duration of an integration cycle.
NOTE
Users logged on in the Service role have write access!

User interface

1 to 65 ms

Measuring period

Navigation

 Application → Sensor → Sensor adjustm. → Measuring period

Description

Displays the duration of a full measuring period.
Additional information:
The measuring period is the time span during which the excitation of the magnetic field
takes place and a measuring point is created.
NOTE
Users logged on in the Service role have write access!

User interface

0 to 1 000 ms

4.4.5"Calibration" submenu

Navigation

 Application → Sensor → Calibration‣ Calibration
Nominal diameter→  92

Calibration factor→  93

Zero point→  93

Conductivity calibration factor→  93Nominal diameter
Navigation

 Application → Sensor → Calibration → Nominal diameter

Description

Shows the nominal diameter of the sensor.

92

Endress+Hauser

Proline Promag 10 HART

User interface"Application" menu

Character string comprising numbers, letters and special characters

Calibration factor
Navigation

 Application → Sensor → Calibration → Cal. factor

Description

Displays the current calibration factor for the flow rate measuring sensor.
Additional information:
The factory setting for the calibration factor can be found on the sensor's nameplate.

User interface

Positive floating-point number

Zero pointNavigation

 Application → Sensor → Calibration → Zero point

Description

Displays the zero point correction value for the sensor.
NOTE
Users logged on in the Service role have write access!

User interface

Signed floating-point number

Conductivity calibration factor



Navigation

 Application → Sensor → Calibration → Cond. cal. fact.

Prerequisite

Conductivity measurement is switched on in the Conductivity measurement parameter(→  86).
Conductivity measurement is only optionally available: Under order code for"Functionality", option D (extended transmitter) and order code for "Sensor option",
option CX (conductivity measurement)

Description

Displays calibration factor for conductivity measurement.
NOTE
Users logged on in the Service role have write access!

User interface

Endress+Hauser

0.01 to 10 000

93

"Application" menu

Proline Promag 10 HART

4.4.6

"Electrode cleaning cycle" submenu

Navigation

 Application → Sensor → Elec. clean cycl

‣ Electrode cleaning cycle
Electrode cleaning cycle→  94ECC duration→  94

ECC recovery time→  94

ECC interval→  95

ECC polarity→  95

Electrode cleaning cycleNavigation

 Application → Sensor → Elec. clean cycl → Elec. clean cycl

Description

Switch electrode cleaning on or off.

Selection

• Off
• On

ECC duration



Navigation

 Application → Sensor → Elec. clean cycl → ECC duration

Description

Enter the duration of the cleaning cycle.

User entry

0.01 to 30 s

ECC recovery timeNavigation

 Application → Sensor → Elec. clean cycl → ECC recov. time

Description

Specify a timespan for recovery after a cleaning cycle has completed to prevent
interference with the signal outputs. The output signal values will be frozen for theduration of the recovery.

User entry

1 to 600 s

94

Endress+Hauser

Proline Promag 10 HART

"Application" menu

ECC interval

Navigation

 Application → Sensor → Elec. clean cycl → ECC interval

Description

Specify the duration of the interval between one cleaning cycle and the next.

User entry

0.5 to 168 h

ECC polarity
Navigation

 Application → Sensor → Elec. clean cycl → ECC polarity

Description

Displays the setting for the electrode cleaning polarity.
Additional information:
The polarity depends on the material of the electrodes:- Platinum: Negative- Tantalum, Alloy C22, stainless steel: Positive

User interface

Endress+Hauser• Positive• Negative

95

"Application" menu

Proline Promag 10 HART

4.5

"Current output" submenu

Navigation

Application → Curr.output 1‣ Current output 1
Process variable current output→  96

Measuring mode current output→  97

Current range output→  101

Fixed current→  102

Lower range value output→  102

Upper range value output→  104

Damping current output→  104

Failure behavior current output→  105

Failure current→  105

Output current 1→  106

Process variable current output
Navigation

 Application → Curr.output 1 → Proc.var. outp

Description

Select process variable for current output

Selection• Off *• Mass flow
• Conductivity *• Corrected conductivity *• Volume flow
• Temperature *• Noise *• Coil current shot time *

*

96



Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Measuring mode current output
Navigation

 Application → Curr.output 1 → Meas.mode outp

Prerequisite

A process variable is selected in the Process variable current output parameter(→  21).

Description

Select the measuring mode for the output.

Selection• Forward flow
• Forward/Reverse flow *• Reverse flow compensation*



Visibility depends on order options or device settings

Endress+Hauser

97

"Application" menu

Additional information

Proline Promag 10 HART

Selection• Forward flow option
The current output signal is proportional to the measured value for the process variableassigned.
Additional information:- The lower limit value ("Lower range value output " parameter) and the upper limit value("Upper range value output " parameter) of the measured value range do not have to havethe same algebraic sign, i.e. the lower limit value can be negative and the upper limit
value positive.
- If the measured value lies outside the scaled measured value range, diagnostic message"441 Current output faulty" is generated.
• Forward/Reverse flow option
The current output outputs the absolute value for the assigned process variable(reflection across the lower limit value of the measured value range).
Additional information:- The lower limit value ("Lower range value output " parameter) and the upper limit value("Upper range value output " parameter) of the measured value range must have thesame algebraic sign.
- If the absolute value exceeds the upper limit value of the measured value range,
diagnostic message "441 Current output faulty" is generated.
- This setting is generally only used for flow-related process variables.
• Reverse flow compensation option
The current output signal is proportional to the measured value for the process variableassigned.
Additional information:- The lower limit value ("Lower range value output " parameter) and the upper limit value("Upper range value output " parameter) of the measured value range do not have to havethe same algebraic sign, i.e. the lower limit value can be negative and the upper limit
value positive.
- Reverse flow (a measured value below the lower limit value of the measured value
range) is stored in a buffer and processed after a maximum delay of 60 s with the next
forward flow.
- When the flow exceeds the maximum value or the reverse flow stored in the buffer
cannot be processed within approx. 60 s, diagnostic message "441 Current output faulty"
is generated.
- This option is used e.g. to compensate intermittent reverse flow, which may occur inconnection with positive displacement pumps as a result of wear and tear or highviscosity.
- There is no flow damping with this setting.
1. Examples of the behavior of the current output
Defined measuring range: lower range value and upper range value with the same
algebraic sign.
I [mA]
20

4
0

1

2

Q
A0028084

I
Q
1
2

98

Current
Flow
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)

Endress+Hauser

Proline Promag 10 HART

"Application" menuWith the following flow response:Qt
A0028091

Q
t

Flow
Time

With the Forward flow option
The current output signal is proportional to the process variable assigned. The flow
components outside the scaled measuring range are not taken into account for signal
output:.
I [mA]

t
A0028092

It

Current
Time

With the Forward/Reverse flow option
The current output signal is independent of the direction of flow.
I [mA]

t
A0028093

It

Current
Time

With the Reverse flow compensation option
Flow components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.

Endress+Hauser

99

"Application" menu

Proline Promag 10 HART

I [mA]At

S
S=A

A0028094

It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

2. Examples of the behavior of the current output
Defined measuring range: lower range value and upper range value with different
algebraic signs.
I [mA]
20
4
0

Q

0
1

2
A0028095

I
Q
1
2

Current
Flow
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With flow a (—) outside, b (- -) inside the measuring range:
Q
1
+

t

0
2

-

a=
b=
A0028098

Q
t
a
b

Flow
Time
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With the Forward flow option• a (—): The flow components outside the scaled measuring range cannot be taken into
account for signal output. The diagnostic message S441 Current output 1 to ndiagnostic message is displayed.
• b (- -): The current output signal is proportional to the process variable assigned.

100

Endress+Hauser

Proline Promag 10 HART

"Application" menu

I [mA]ta=b=
A0028100It

Current
Time

With the Forward/Reverse flow option
This option cannot be selected here since the values for the 20 mA value parameter(→  23) and 20 mA value parameter (→  23) have different algebraic signs.
With the Reverse flow compensation option
Flow components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.
I [mA]At

S
S=A

A0028101It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

Current range output

Navigation

 Application → Curr.output 1 → Curr.range out

Description

Select the current range for the measured value output and the upper and lower fault
condition signal level.
Additional information:- The measured value range is specified in the "Lower range value output " parameter andthe "Upper range value output " parameter.
- If the measured value lies outside the scaled measured value range, diagnostic message"441 Current output faulty" is generated.
- In the event of a device alarm, the current output adopts the behavior specified in the"Failure behavior current output " parameter.

Selection

Endress+Hauser• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)• Fixed value

101

"Application" menu

Additional information

Proline Promag 10 HART

Selection• 4...20 mA NE (3.8...20.5 mA) option
Select this option to set the current range in accordance with NAMUR recommendation
NE43.
• Fixed value option
Select this option to set the current output to a current value instead of a range.
The current value is defined in the Fixed current parameter (→  24).
The graphic shows the relationship between the current range for the output of theprocess value and the lower and upper alarm levels:
2

1

3

I [mA]
A0034351

1
2
3

Current range for process value
Lower level for signal on alarm
Upper level for signal on alarm

Selection (current range for process value) Lower level for signal on
alarm

Upper level for signal on
alarm

4...20 mA NE (3.8...20.5 mA)> 21.5 mA< 3.6 mA

4...20 mA US (3.9...20.8 mA)
4...20 mA (4... 20.5 mA)

Fixed current

Navigation

 Application → Curr.output 1 → Fixed current

Prerequisite

In the Current range output parameter in the Current output 1 submenu, the Fixedvalue option is selected.

Description

Enter the value for the "Fixed value" option.

User entry

3.59 to 21.5 mA

Lower range value output

Navigation

 Application → Curr.output 1 → Low.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

102

Endress+Hauser

Proline Promag 10 HART

Description"Application" menuEnter lower range value for the measured value range.
Additional information:- Depending on the setting selected for the "Measuring mode current output " parameter,
the value specified for this parameter and the "Upper range value output " parameter musthave the same algebraic sign or not.
- As a rule, the lower range value is scaled to be lower than the upper range value. As aresult, the behavior of the current output is proportional to the process variable assigned.
If the lower range value is scaled to be higher than the upper range value, then the
behavior of the current output will be inversely proportional to the process variableassigend.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter (→  97).
Example: Measuring mode with "Forward flow" option• Lower range value output parameter (→  22) = not equal to zero flow (e.g. –250
m3/h)
• Upper range value output parameter (→  23) = not equal to zero flow (e.g. +750
m3/h)
• Calculated current value = 8 mA at zero flow
I [mA]
20
84Q-250 01+750
1A0013757

Q
I
1

Flow
Current
Measuring range is exceeded or undershot

The linear operational range of the measuring device is defined by the values entered for
the Lower range value output parameter (→  22) and Upper range value output
parameter (→  23), and by the selected current range.
Example: Measuring mode with the "Forward/Reverse flow" option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

Q
I
1
2
3

Endress+Hauser

Flow
Current
Value assigned to 0/4 mA current
Forward flow
Reverse flow

103

"Application" menu

Proline Promag 10 HART

The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the Lower range value output parameter (→  22)
and Upper range value output parameter (→  23) must have the same algebraic sign.
The value for the Upper range value output parameter (→  23) (e.g. reverse flow)corresponds to the mirrored value for the Upper range value output parameter (→  23)(e.g. forward flow).
Example: Measuring mode with the "Reverse flow compensation" option
If flow is characterized by severe fluctuations (e.g. when using reciprocating pumps), flow
components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s.

Upper range value output

Navigation

 Application → Curr.output 1 → Upp.range outp

Prerequisite

In the Current range output parameter, one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter upper range value for the measured value range.

User entry

Signed floating-point number

Additional information

Examples of the behavior, depending on the option selected in the Measuring modecurrent output parameter: Lower range value output parameter (→  22):

Damping current output

Navigation

 Application → Curr.output 1 → Damp.curr.outp

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Enter time constant (PT1 element) to set the reaction time of the output signal tofluctuations in the measured value caused by process conditions.
Additional information:- The smaller the time constant the faster the output reacts to fluctuations in themeasured value.
- If the time constant is 0, damping is deactivated.

User entry

104

0.0 to 999.9 s

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Failure behavior current output

Navigation

 Application → Curr.output 1 → Fail.behav.out

Prerequisite

A process variable is selected in the Process variable current output parameter and oneof the following options is selected in the Current range output parameter:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4... 20.5 mA)

Description

Specify how the output should behave in the event of a device alarm.

Selection• Min.
• Max.
• Last valid value• Actual value• Fixed value

Additional information

Selection• Min. option
The current output outputs the lower fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Max. option
The current output outputs the upper fault condition signal level for the specified current
range.
Additional information:
The current range is specified in the "Current range output " parameter.
• Last valid value option
The current output outputs the last valid value measured before the device alarm
occurred.
• Actual value option
The current output outputs the flow value currently measured. The device alarm isignored.
• Fixed value option
The current output outputs the value specified.
Additional information:
The value is specified in the "Failure current " parameter.

Failure current

Navigation

 Application → Curr.output 1 → Fail. current

Prerequisite

In the Failure behavior current output parameter in the Current output 1 submenu, theFixed value option is selected.

Description

Enter the value for the "Fixed value" option in the "Failure behavior current output "parameter.

User entry

3.59 to 21.5 mA

Endress+Hauser

105

"Application" menu

Proline Promag 10 HART

Output current
Navigation

 Application → Curr.output 1 → Output curr. 1

Description

Displays the current value currently calculated.

User interface

3.59 to 21.5 mA

106Endress+Hauser

Proline Promag 10 HART

"Application" menu

4.6

"Pulse/frequency/switch output 1" submenu

Configuring the pulse/frequency/switch output
Navigation

 Application → PFS output 1‣ Pulse/frequency/switch output 1

Endress+Hauser

Operating mode→  108

Assign pulse output 1→  110

Measuring mode→  111

Value per pulse→  111

Pulse width→  112

Failure mode→  112

Pulse output 1→  113

Assign frequency output→  114

Measuring mode→  114Minimum frequency value→  115

Maximum frequency value→  116

Measuring value at minimum
frequency→  116

Measuring value at maximum
frequency→  116

Damping output 1→  117

Failure mode→  117

Failure frequency→  117

Output frequency 1→  118

Switch output function→  118

Assign diagnostic behavior→  119

Assign limit→  119107

"Application" menu

Proline Promag 10 HART

Switch-on value→  121

Switch-off value→  122

Switch-on delay→  122

Switch-off delay→  122

Assign flow direction check→  122

Assign status→  123

Failure mode→  123

Invert output signal→  124

Switch state 1→  124

Operating modeNavigation

 Application → PFS output 1 → Operating mode

Description

Select the operating mode for the output.

Selection• Pulse• Frequency• Switch

108

Endress+Hauser

Proline Promag 10 HART

Additional information"Application" menu

Selection• Pulse option
Quantitatively proportional pulse with pulse width to be configured. Whenever the pulse
value for the specified process variable has been reached, a pulse is emitted, the durationof which is set within the "Pulse width" parameter.
Additional information:
The process variable for the pulse output is specified in the "Assign pulse output "parameter.
• Frequency option
The output frequency is proportional to the value for the process variable assigned, witha pulse-to-interval ratio of 1:1.
Additional information:
The process variable for the frequency output is specified in the "Assign frequency
output" parameter.
• Switch option
Indicates when the state of the device changes, e.g. when a specified limit value isreached or an alarm or warning is triggered.
Additional information:- The switch output can be in one of two states: either it is conductive or it is nonconductive.
- When the function assigned to the switch output is triggered, the switch output will
depending on the output configuration either be continuously conductive or continuously
non-conductive.
- The switch output is used to display diagnostic information at the system level, e.g. byconnecting a lamp that lights up when the function assigned is triggered.
"Pulse" option
Example• Flow rate approx. 100 g/s• Pulse value 0.1 g• Pulse width 0.05 ms• Pulse rate 1 000 pulse/s
U[V]

B<P
B

24
t

0
P

A0026883

5
BP

Quantity-proportional pulse (pulse value) with pulse width to be configured
Pulse width entered
Pauses between the individual pulses"Frequency" option
Example• Flow rate Q approx. 100 g/s• Min. frequency (fmin) 0 Hz
• Max. frequency (fmax) 1000 Hz• Flow rate at min. frequency (Qmin) 0 g/s• Flow rate at max. frequency (Qmax) 1000 g/s• Output frequency (fout) approx. 100 Hz
fout = fmin + Q × [(fmax - fmin)/(Qmax - Qmin)] =0 Hz + 100 g/s × [(1000 Hz - 0 Hz)/(1000 g/s - 0 g/s )] = 100 Hz
Endress+Hauser

109

"Application" menu

Proline Promag 10 HART

U[V]

fout
T!T"

T!= T"

24t

0

A0026886

6

Flow-proportional frequency output"Switch" option
ExampleAlarm response without alarm
U[V]
24
t

0

A0026884

7

No alarm, high level

ExampleAlarm response in case of alarm
U[V]
24
0

t
A0026885

8

Alarm, low level

Assign pulse output 1



Navigation

 Application → PFS output 1 → Assign pulse 1

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Select process variable for pulse output.

Selection• Off• Volume flow
• Mass flow

110

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Measuring modeNavigation

 Application → PFS output 1 → Measuring mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Select measuring mode for pulse output.

Selection• Forward flow
• Forward/Reverse flow
• Reverse flow
• Reverse flow compensation

Additional information

Selection• Forward flow option
Positive flow is output, negative flow not output.
• Forward/Reverse flow option
Both positive and negative flow are output (absolute value), whereby no distinction ismade between positive and negative flow.
• Reverse flow option
Negative flow is output, positive flow is not output.
• Reverse flow compensation option
Positive flow is output. Negative flow quantities are buffered, processed, and output after
a maximum delay of 60 s.
Additional information:
This option is used e.g. to compensate intermittent negative flow, which may occur inconnection with positive displacement pumps as a result of wear and tear or highviscosity.

Value per pulseNavigation

 Application → PFS output 1 → Value per pulse

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Enter the measured value to which a pulse corresponds.
Additional information:
Weighting of the pulse output with a quantity.
The lower the pulse value, the– better the resolution.
– higher the frequency of the pulse response.

User entry

Endress+Hauser

Signed floating-point number

111

"Application" menu

Proline Promag 10 HART

Pulse widthNavigation

 Application → PFS output 1 → Pulse width

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Specify the duration of the output pulse.
Additional information:
The maximum pulse rate is defined by fmax = 1 / (2 × pulse width). The interval between
two pulses (P) is at least as long as the specified pulse width (B).
The maximum flow is defined by Qmax = fmax × pulse value. If the flow exceeds theselimit values, the measuring device displays the diagnostic message "443 Pulse output
faulty".
Example:- Pulse value: 0.1 g- Pulse width: 0.1 ms- fmax: 1 / (2 × 0.1 ms) = 5 kHz- Qmax: 5 kHz × 0.1 g = 0.5 kg/s

User entry

0.05 to 2 000 ms

Additional information

Description
U[V]

U[V]

B<P

B=P
B

B

t
P

t
P
A0026882

B
P

Pulse width entered
Pauses between the individual pulses

Failure modeNavigation

 Application → PFS output 1 → Failure mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Specify how the output should behave in the event of a device alarm.
Additional information:
For safety reasons, it is recommended that the behavior of the output in the event of adevice alarm be predefined.

Selection

112• Actual value• No pulses

Endress+Hauser

Proline Promag 10 HART

Additional information"Application" menu

Selection• Actual value option
In the event of a device alarm, the pulse output continues based on the current flow
measurement. The issue is ignored.
Additional information:
A device alarm indicates a serious malfunction of the measuring device that may impact
the measurement quality to the point that accuracy can no longer be ensured. This
option is only recommended if the necessary safeguards are in place to ensure that noalarm condition can impact the measurement quality.
• No pulses option
In the event of a device alarm, no pulses are emitted.

Pulse output 1
Navigation

 Application → PFS output 1 → Pulse output 1

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, thePulse option is selected.

Description

Displays the pulse frequency currently output.
Additional information:- The output behavior can be inverted in the "Invert output signal" parameter, i.e. in thiscase the transistor will be non-conductive for the duration of a pulse.
- The "Invert output signal" parameter is not available for all devices.

User interface

Positive floating-point number

Additional information

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.

1

1

t

0
NC

t

0
NO

A0028726

0
1
NC
NO

Non-conductive
Conductive
Normally closed
Normally open

The behavior of the output in the event of a device alarm can be configured: Failuremode parameter (→  112)

Endress+Hauser

113

"Application" menu

Proline Promag 10 HART

Assign frequency output

Navigation

 Application → PFS output 1 → Assign freq.

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Select process variable for frequency output.

Selection• Off• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Noise *• Coil current shot time *Measuring modeNavigation

 Application → PFS output 1 → Measuring mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Select measuring mode for frequency output.

Selection• Forward flow
• Forward/Reverse flow
• Reverse flow compensation*

114

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

Additional information"Application" menu

Selection• Forward flow option
The output signal is proportional to the measured value for the process variableassigned.
Additional information:- The lower limit value ("Measuring value at minimum frequency" parameter) and theupper limit value ("Measuring value at maximum frequency" parameter) of the measured
value range do not have to have the same algebraic sign, i.e. the lower limit value can be
negative and the upper limit value positive.
- If the measured value lies outside the scaled measured value range, diagnostic message"442 Frequency output faulty" is generated.
• Forward/Reverse flow option
The output outputs the absolute value for the assigned process variable (reflection acrossthe lower limit value of the measured value range).
Additional information:- The lower limit value ("Measuring value at minimum frequency" parameter) and theupper limit value ("Measuring value at maximum frequency" parameter) of the measured
value range must have the same algebraic sign.
- If the absolute value exceeds the upper limit value of the measured value range,
diagnostic message "442 Frequency output faulty" is generated.
- This setting is generally used for flow-related process variables.
• Reverse flow compensation option
The output signal is proportional to the measured value for the process variableassigned.
Additional information:- The lower limit value ("Measuring value at minimum frequency" parameter) and theupper limit value ("Measuring value at maximum frequency" parameter) of the measured
value range do not have to have the same algebraic sign, i.e. the lower limit value can be
negative and the upper limit value positive.
- Reverse flow (a measured value below the lower limit value of the measured value
range) is stored in a buffer and processed after a maximum delay of 60 s with the next
forward flow.
- When the flow exceeds the maximum value or the reverse flow stored in the buffer
cannot be processed within approx. 60 s, diagnostic message "442 Frequency output
faulty" is generated.
- This option is used e.g. to compensate intermittent reverse flow, which may occur inconnection with positive displacement pumps as a result of wear and tear or highviscosity.
- There is no flow damping with this setting.

Minimum frequency value

Navigation

 Application → PFS output 1 → Min. freq. value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter the minimum frequency for the frequency range.
Additional information:
The lower range value for the measured value range is specified in the "Measuring value atminimum frequency" parameter.

User entry

Endress+Hauser

0.0 to 10 000.0 Hz

115

"Application" menu

Proline Promag 10 HART

Maximum frequency value

Navigation

 Application → PFS output 1 → Max. freq. value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter the maximum frequency for the measured value output.
Additional information:
The upper range value for the measured value range that corresponds to the maximum
frequency is specified in the "Measuring value at maximum frequency" parameter.

User entry

0.0 to 10 000.0 Hz

Measuring value at minimum frequency

Navigation

 Application → PFS output 1 → Val. at min.freq

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter lower range value for the measured value range.
Additional information:- Depending on the setting selected for the "Measuring mode" parameter, the value
specified for this parameter and the "Measuring value at maximum frequency" parametermust have the same algebraic sign or not.
- As a rule, the lower range value is scaled to be lower than the upper range value. As aresult, the behavior of the frequency output is proportional to the process variableassigned. If the lower range value is scaled to be higher than the upper range value, thenthe behavior of the frequency output will be inversely proportional to the process variableassigend.

User entry

Signed floating-point number

Measuring value at maximum frequency

Navigation

 Application → PFS output 1 → Val. at max.freq

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter upper range value for the measured value range.

User entry

Signed floating-point number

116

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Damping output 1



Navigation

 Application → PFS output 1 → Damping out. 1

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Enter time constant (PT1 element) to set the reaction time of the output signal tofluctuations in the measured value caused by process conditions.
Additional information:- The smaller the time constant the faster the output reacts to fluctuations in themeasured value.
- If the time constant is 0, damping is deactivated.

User entry

0 to 999.9 s

Failure modeNavigation

 Application → PFS output 1 → Failure mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Specify how the output should behave in the event of a device alarm.
Additional information:
For safety reasons, it is recommended that the behavior of the output in the event of adevice alarm be predefined.

Selection• Actual value• Defined value• 0 Hz

Additional information

Selection• Actual value option
The frequency output outputs the flow value currently measured.
• Defined value option
The frequency output outputs the value specified.
Additional information:
The value is specified in the "Failure frequency" parameter.
• 0 Hz option
In the event of a device alarm, the frequency output outputs 0 Hz.

Failure frequency

Navigation

 Application → PFS output 1 → Failure freq.

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Endress+Hauser

117

"Application" menu

Proline Promag 10 HART

Description

Enter the value for the "Defined value" option in the "Failure mode" parameter.

User entry

0.0 to 12 500.0 Hz

Output frequency 1
Navigation

 Application → PFS output 1 → Output freq. 1

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theFrequency option is selected.

Description

Displays the pulse frequency currently output.

User interface

0.0 to 12 500.0 Hz

Switch output function



Navigation

 Application → PFS output 1 → Switch out funct

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Assign a function to the switch output.
Additional information:- The state of the switch output (on or off) when the assigned function is triggered can beinverted in the "Invert output signal" parameter- The "Invert output signal" parameter is not available for all devices.

Selection

• Off
• On
• Diagnostic behavior• Limit• Flow direction check• Status

Additional information

Selection• Off option
The switch output is permanently switched off (open, non-conductive).
• On option
The switch output is permanently switched on (closed, conductive).
• Diagnostic behavior option
The switch output is switched on (closed, conductive), if there is a pending diagnostic
event of the assigned behavioral category.

118

Endress+Hauser

Proline Promag 10 HART

"Application" menu

• Limit option
The switch output is switched on (closed, conductive), if a limit value specified for theprocess variable is reached.
• Flow direction check option
The switch output is switched on (closed, conductive), when the flow direction changes(forward or reverse flow).
• Status option
The switch output is switched on (closed, conductive) to indicate the device status for theselected detection method, e.g. empty pipe detection or low flow cut off.

Assign diagnostic behavior

Navigation

 Application → PFS output 1 → Assign diag. beh

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

The switch output is switched on (closed, conductive), if there is a pending diagnostic
event of the assigned behavioral category.

Selection• Alarm• Alarm or warning• Warning

Additional information

Selection• Alarm option
The switch output is only switched on for diagnostic events of the "Alarm" category.
• Alarm or warning option
The switch output is switched on for diagnostic events of the "Alarm" or "Warning"
category.
• Warning option
The switch output is only switched on for diagnostic events of the "Warning" category.

Assign limitNavigation

 Application → PFS output 1 → Assign limit

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Select the process variable to monitor in case the specified limit value is exceeded. If a limit
value for the selected process variable is exceeded, the output is switched on.

Selection• Off• Volume flow
• Mass flow
• Flow velocity• Conductivity **Visibility depends on order options or device settings

Endress+Hauser

119"Application" menu

Proline Promag 10 HART

• Corrected conductivity *• Totalizer 1• Totalizer 2• Totalizer 3• Temperature *• Sensor electronics temperature (ISEM)
Additional information

Switch-on point > switch-off point
Behavior of the status output if switch-on point > switch-off point:• Process variable > switch-on point: transistor is conductive• Process variable < switch-off point: transistor is not conductive
A
1
2
t
B
3
4

t
A0026891

1
2
3
4
A
B

Switch-on point
Switch-off point
Conductive
Non-conductive
Process variable
Status output

Switch-on point < switch-off point
Behavior of the status output if switch-on point < switch-off point:• Process variable < switch-on point: transistor is conductive• Process variable > switch-off point: transistor is not conductive*

120

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"Application" menu

A
1
2
t
B
3
4

t
A0026892

1
2
3
4
A
B

Switch-on point
Switch-off point
Conductive
Non-conductive
Process variable
Status output

Switch-on point = switch-off point
Behavior of the status output if switch-on point = switch-off point:• Process variable > switch-on point: transistor is conductive• Process variable < switch-off point: transistor is not conductive
A
1
t
B
2
3

t
A0026893

1
2
3
A
B

Switch-on point = switch-off point
Conductive
Non-conductive
Process variable
Status output

Switch-on value

Navigation

 Application → PFS output 1 → Switch-on value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter limit value for the switch-on point (process variable > switch-on value = closed,
conductive).
Additional information:
To use a hysteresis: Switch-on point > Switch-off point.

User entry

Endress+Hauser

Signed floating-point number

121

"Application" menu

Proline Promag 10 HART

Switch-off value

Navigation

 Application → PFS output 1 → Switch-off value

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter limit value for the switch-off point (process variable < switch-off value = open, nonconductive).
Additional information:
To use a hysteresis: Switch-on point > Switch-off point.

User entry

Signed floating-point number

Switch-on delay

Navigation

 Application → PFS output 1 → Switch-on delay

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter delay before the switch output is switched on.

User entry

0.0 to 100.0 s

Switch-off delay

Navigation

 Application → PFS output 1 → Switch-off delay

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Enter delay before the switch output is switched off.

User entry

0.0 to 100.0 s

Assign flow direction checkNavigation

 Application → PFS output 1 → Assign dir.check

Prerequisite

In the Switch output function parameter in the Pulse/frequency/switch output 1
submenu, the Frequency option is selected.

Description

Select process variable for flow direction monitoring.

122

Endress+Hauser

Proline Promag 10 HART

Selection"Application" menu

• Off
• Volume flow
• Mass flow

Assign status



Navigation

 Application → PFS output 1 → Assign status

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Select the device status to display for the switch output.
Additional information:
If the switch on point for the selected detection method is reached, the output isconductive. Otherwise, the switch output is non-conductive.

Selection• Empty pipe detection• Low flow cut off

Failure modeNavigation

 Application → PFS output 1 → Failure mode

Prerequisite

In the Operating mode parameter in the Pulse/frequency/switch output 1 submenu, theSwitch option is selected.

Description

Specify how the output should behave in the event of a device alarm.
Additional information:
For safety reasons, it is recommended that the behavior of the output in the event of adevice alarm be predefined.

Selection• Actual status• Open• Closed

Additional information

Selection• Actual status option
In the event of a device alarm, the issue is ignored and the switch output adopts the
behavior currently specified for the "Switch output function" parameter.
• Open option
In the event of a device alarm, the switch output's transistor is set to "non-conductive".

Endress+Hauser

123

"Application" menu

Proline Promag 10 HART

Invert output signal

Navigation

 Application → PFS output 1 → Invert outp.sig.

Description

Indicate whether to invert the output signal (Yes/No).
Additional information:- If the output signal is inverted, the output behavior is the reverse of its configuration.
- This setting does not apply to the frequency output.

Selection

• No
• Yes

Additional information

Selection: no (passive - negative)

1
0
A0026693

Selection: yes (passive - positive)

1
0
A0026692

Switch state 1
Navigation

 Application → PFS output 1 → Switch state 1

Description

Indicates the current switch output status.

User interface• Open• Closed

Additional information

User interface• Open option
The switch output is not conductive.
• Closed option
The switch output is conductive.

124

Endress+Hauser

Proline Promag 10 HART

"Application" menu

4.7"Communication" submenu

Navigation

 Application → Communication‣ Communication

4.7.1
Navigation‣ HART configuration→  125‣ Information→  126‣ Output→  128"HART configuration" submenu
 Application → Communication → HART config.

‣ HART configuration
HART short tag→  125

Device tag→  125

HART address→  126

HART short tag

Navigation

 Application → Communication → HART config. → HART short tag

Description

Enter a brief description for the measuring point.

User entry

Character string comprising numbers, letters and special characters (8)Device tagNavigation

 Application → Communication → HART config. → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)Endress+Hauser

125

"Application" menu

Proline Promag 10 HART

HART addressNavigation

 Application → Communication → HART config. → HART address

Description

Enter the address to exchange data via the HART protocol.

User entry

0 to 63

4.7.2"Information" submenu

Navigation

 Application → Communication → Information‣ Information
Device revision→  126

Device ID→  127

Device type→  127

Manufacturer ID→  127

HART revision→  127

HART descriptor→  127

HART message→  128

Hardware revision→  128

Software revision→  128

HART date code→  128

Device revision
Navigation

 Application → Communication → Information → Device revision

Description

Displays device revision.

User interface

0 to 255

126

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Device ID

Navigation

 Application → Communication → Information → Device ID

Description

Displays the device ID to identify the measuring device.

User interface

Positive integer

Device type
Navigation

 Application → Communication → Information → Device type

Description

Displays the device type registered with the HART Communication Foundation.

User interface

0 to 65 535

Manufacturer ID
Navigation

 Application → Communication → Information → Manufacturer ID

Description

Displays the manufacturer ID registered with the HART Communication Foundation.

User interface

0 to 65 535

HART revision
Navigation

 Application → Communication → Information → HART revision

Description

Displays the HART protocol revision of the measuring device.

User interface

5 to 7

HART descriptor

Navigation

 Application → Communication → Information → HART descriptor

Description

Enter description for the measuring point

User entry

Character string comprising numbers, letters and special characters (16)Endress+Hauser

127

"Application" menu

Proline Promag 10 HART

HART message

Navigation

 Application → Communication → Information → HART message

Description

Enter HART message to be sent via HART protocol when requested by the master.

User entry

Character string comprising numbers, letters and special characters (32)

Hardware revision
Navigation

 Application → Communication → Information → Hardware rev.

Description

Displays the hardware revision of the measuring device.

User interface

0 to 255

Software revision
Navigation

 Application → Communication → Information → Software rev.

Description

Displays software revision of the measuring device.

User interface

0 to 255

HART date code

Navigation

 Application → Communication → Information → HART date code

Description

Enter date for individual use.

User entry

Character string comprising numbers, letters and special characters (10)4.7.3
Navigation"Output" submenu
 Application → Communication → Output‣ Output
Assign PV128→  129

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Primary variable (PV)→  129

Assign SV→  130

Secondary variable (SV)→  130

Assign TV→  130

Tertiary variable (TV)→  131

Assign QV→  131

Quaternary variable (QV)→  131

Assign PV

Navigation

 Application → Communication → Output → Assign PV

Description

Assign a measured variable to the primary dynamic variable (PV).
Additional information:
The assigned measured variable is also used by the current output.

Selection• Off *• Mass flow
• Conductivity *• Corrected conductivity *• Volume flow
• Temperature *• Noise *• Coil current shot time *Primary variable (PV)Navigation

 Application → Communication → Output → Primary var (PV)

Description

Displays the value currently measured for the primary dynamic variable.

User interface

Signed floating-point number*

Visibility depends on order options or device settings

Endress+Hauser

129

"Application" menu

Proline Promag 10 HART

Assign SVNavigation

 Application → Communication → Output → Assign SV

Description

Assign a measured variable to the second dynamic variable (SV).

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Noise *• Coil current shot time *• Totalizer 1• Totalizer 2• Totalizer 3

Secondary variable (SV)Navigation

 Application → Communication → Output → Second.var(SV)Description

Displays the value currently measured for the secondary dynamic variable.

User interface

Signed floating-point number

Assign TV

Navigation

 Application → Communication → Output → Assign TV

Description

Assign a measured variable to the tertiary dynamic variable (TV).

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Noise *• Coil current shot time *• Totalizer 1• Totalizer 2• Totalizer 3*

130

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"Application" menu

Tertiary variable (TV)
Navigation

 Application → Communication → Output → Tertiary var(TV)Description

Displays the value currently measured for the tertiary dynamic variable.

User interface

Signed floating-point number

Assign QVNavigation

 Application → Communication → Output → Assign QV

Description

Assign a measured variable to the quaternary dynamic variable (QV).

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Noise *• Coil current shot time *• Totalizer 1• Totalizer 2• Totalizer 3

Quaternary variable (QV)
Navigation

 Application → Communication → Output → Quaterna.var(QV)

Description

Displays the value currently measured for the quaternary dynamic variable.

User interface

Signed floating-point number*

Visibility depends on order options or device settings

Endress+Hauser

131

"System" menu

Proline Promag 10 HART

5

"System" menu

Overall device management and security settings – management of system settings andadaption to operational requirements.
Navigation

 System

System132‣ Device management→  133‣ User management→  135‣ Connectivity→  138‣ Date/time→  139‣ Information→  141‣ Display→  146‣ Software configuration→  150

Endress+Hauser

Proline Promag 10 HART

"System" menu

5.1

"Device management" submenu

Navigation

 System → Device manag.

‣ Device management
Device tag→  133

Locking status→  133

Configuration counter→  134

Device reset→  134

Device tagNavigation

 System → Device manag. → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)

Locking status
Navigation

 System → Device manag. → Locking status

Description

Indicates the write protection with the highest priority that is currently active.

User interface• Hardware locked• Temporarily locked

Additional information

User interface• Hardware locked option
The DIP switch for the hardware lock is enabled. As a result write access to theparameters is locked.
• Temporarily locked option
Due to internal procedures that are currently in progress (e.g. data upload/download,
reset, etc.), write access to the parameters is temporarily locked.The parameters can be
modified again, once the internal procedures are complete.

Endress+Hauser

133

"System" menu

Proline Promag 10 HART

Configuration counter
Navigation

 System → Device manag. → Config. counter

Description

Displays the counter for changes to the device parameters.
Additional information:- If the value for a static parameter is changed when optimizing or configuring theparameter, the counter is incremented by 1. This is to enable tracking different parameterversions.
- When multiple parameters are changed simultaneously, e.g. when loading parameters
into the device from an external source such as FieldCare, the counter may display a higher
value. The counter cannot be reset, nor is it reset to a default value on performing a device
reset.
- Once the counter has reached the value 65535, it restarts at 0.

User interface

0 to 65 535

Device reset

Navigation

 System → Device manag. → Device reset

Description

Reset the device configuration - either entirely or in part - to a defined state.

Selection• Cancel• To delivery settings• Restart device• Restore S-DAT backup *• Create T-DAT backup• Restore T-DAT backup *Additional information

Selection• To delivery settings option
Every parameter for which a customer-specific default setting was ordered is reset to thecustomer-specific value. All other parameters are reset to the factory setting.
• Restart device option
The restart resets every parameter with data stored in volatile memory (RAM) to thefactory setting (e.g. measured value data). The device configuration remains unchanged.
• Restore S-DAT backup option
Restore the data that is saved on the S-DAT. The data record is restored from theelectronics memory to the S-DAT.
• Create T-DAT backup option
Create T-DAT backup.

*

134

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"System" menu

5.2

"User management" submenu

Navigation

 System → User manag.

‣ User management
User role→  135

Enter access code→  136

Reset Maintenance code→  136‣ Define Maintenance code→  137

User role
Navigation

 System → User manag. → User role

Description

Displays the role the user is currently logged on in. The role determines the user's accessrights for the parameters.
Additional information:- Until a Maintenance code has been set in the "Define Maintenance code" parameter, allusers are automatically logged on in the Maintenance role. Once the Maintenance code
has been set, all users are automatically logged on in the Operator role.
- The access rights can be changed via the "Enter access code" parameter.

User interface• Operator• Maintenance• Service• Production• Development

Additional information

User interface• Operator option
Provides only read access to parameters.
• Maintenance option
Provides read and write access to parameters.
Additional information:
For some parameters, the user must be logged on in the Service role to obtain write
access.
• Service option
Provides read and write access to Service parameters.

Endress+Hauser

135

"System" menu

Proline Promag 10 HARTEnter access code
Navigation

 System → User manag. → Ent. access code

Description

For users logged on in the Operator role, enter the Maintenance code to change the accessstatus to Maintenance and disable write protection of parameters. For users logged on in
the Maintenance role, enter the Service code to change the access status to Service andenable read and write access to Service parameters.

User entry

0 to 9 999

Reset Maintenance code
Navigation

 System → User manag. → Reset Maint code

Description

Enter the code provided by Endress+Hauser Technical Support to reset the Maintenancecode.

User entry

Character string comprising numbers, letters and special characters (32)136

Endress+Hauser

Proline Promag 10 HART

"System" menu

5.2.1

"Define access code" wizard

Complete this wizard to specify an access code for the Maintenance role.
Navigation

 System → User manag. → Def. access code

‣ Define Maintenance code
Define Maintenance code→  137

Confirm Maintenance code→  137

Define Maintenance code

Navigation

 System → User manag. → Def. Maint. code → Def. Maint. code

Description

Specify an access code that is required to obtain the access rights for the Maintenance role.

User entry

0 to 9 999

Confirm Maintenance code

Navigation

 System → User manag. → Def. Maint. code → Conf. Maint code

Description

Confirm the access code entered for the Maintenance role.

User entry

0 to 9 999

Endress+Hauser

137

"System" menu

Proline Promag 10 HART

5.3

"Connectivity" submenu

Navigation

 System → Connectivity‣ Connectivity‣ Bluetooth configuration

5.3.1→  138"Bluetooth configuration" submenu

Navigation

 System → Connectivity → Bluetooth conf.

‣ Bluetooth configuration
Bluetooth→  138

Communication established→  138

BluetoothNavigation

 System → Connectivity → Bluetooth conf. → Bluetooth

Description

Enable or disable Bluetooth.

Selection• Enable• Disable• Not available *Communication established
Navigation

 System → Connectivity → Bluetooth conf. → Communi. establ.

User interface

• No
• Yes

*

138

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"System" menu

5.4

" Date / Time" submenu

Navigation

 System → Date / Time‣ Date/time
Set date/time→  139

Time format→  139

Time zone→  139

Set date/timeNavigation



System → Date/time → Set date/time

Description

Set the date and local time. Every time the date or time is changed, a logbook entry iscreated.

User entry

Date and time

Time format



Navigation

 System → Date/time → Time format

Description

Select time format.

Selection• 24 h• 12 h AM/PMTime zoneNavigation

 System → Date/time → Time zone

Description

Select the time zone. Every time the time zone is changed, a logbook entry is created.

Endress+Hauser

139

"System" menu

Selection

140

Proline Promag 10 HART

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+14:00

Endress+Hauser

Proline Promag 10 HART

"System" menu

5.5

"Information" submenu

Navigation

 System → Information‣ Information

5.5.1
Navigation‣ Device→  141‣ Sensor electronic module (ISEM)→  144‣ Display module→  144"Device" submenu
 System → Information → Device‣ Device
Device name→  141

Device tag→  142

Serial number→  142

Order code→  142

Firmware version→  142

Extended order code 1→  143

Extended order code 2→  143

Extended order code 3→  143

ENP version→  143

Manufacturer→  144

Device name
Navigation

 System → Information → Device → Device name

Description

Displays the name of the transmitter.
Additional information:
The name can also be found on the transmitter's nameplate.

Endress+Hauser

141

"System" menu

User interface

Proline Promag 10 HART

Character string comprising numbers, letters and special characters

Device tag
Navigation

 System → Information → Device → Device tag

Description

Displays the name for the measuring point.

User interface

Character string comprising numbers, letters and special characters

Serial number
Navigation

 System → Information → Device → Serial number

Description

Displays the serial number of the measuring device. The serial number can be used to
identify the measuring device and to retrieve further information on the measuring device,
such as the related documentation, via the Device Viewer or Operations app.
Additional information:
The serial number can also be found on the nameplate of the sensor and transmitter.

User interface

Character string comprising numbers, letters and special characters

Order code

Navigation

 System → Information → Device → Order code

Description

Displays the device order code.
Additional information:
The order code can be used for instance to order a replacement or spare device or to verify
that the device features specified on the order form match the shipping note.

User interface

Character string comprising numbers, letters and special characters

Firmware version
Navigation

 System → Information → Device → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

142

Endress+Hauser

Proline Promag 10 HART

"System" menu

Extended order code 1Navigation

 System → Information → Device → Ext. order cd. 1

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Extended order code 2Navigation

 System → Information → Device → Ext. order cd. 2

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Extended order code 3

Navigation

 System → Information → Device → Ext. order cd. 3

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

ENP version
Navigation

 System → Information → Device → ENP version

Description

Displays the version of the electronic nameplate (ENP).

Endress+Hauser

143

"System" menu

User interface

Proline Promag 10 HART

Character string comprising numbers, letters and special characters

Manufacturer
Navigation

 System → Information → Device → Manufacturer

Description

Displays the manufacturer.

User interface

Character string comprising numbers, letters and special characters

5.5.2"Sensor electronic module (ISEM)" submenu

Navigation

 System → Information → Sens. electronic‣ Sensor electronic module (ISEM)
Firmware version→  144

Firmware version
Navigation

 System → Information → Sens. electronic → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

5.5.3"Display module" submenu

Navigation

 System → Information → Display module‣ Display module
Firmware version144→  145

Endress+Hauser

Proline Promag 10 HART

"System" menu

Firmware version
Navigation

 System → Information → Display module → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

Endress+Hauser

145

"System" menu

Proline Promag 10 HART

5.6

"Display" submenu

Navigation

 System → Display‣ Display
Language→  146

Value 1 display→  147

Value 2 display→  147

Value 3 display→  148

Value 4 display→  148

Display damping→  148

Rotation display→  149

Brightness→  149

Color scheme→  149

Language
Navigation

 System → Display → Language

Description

Set display language.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• ‫( الْعَرَبيّة‬Arabic) *• Bahasa Indonesia*

146

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

"System" menu• ภาษาไทย (Thai) *• tiếng Việt (Vietnamese)• čeština (Czech)

Value 1 display

Navigation

 System → Display → Value 1 display

Description

Select the measured value that is displayed first on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 2 display

Navigation

 System → Display → Value 2 display

Description

Select the measured value that is shown second on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection*• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Visibility depends on order options or device settings

Endress+Hauser

147

"System" menu

Proline Promag 10 HART

Value 3 display

Navigation

 System → Display → Value 3 display

Description

Select the measured value that is shown third on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Value 4 display

Navigation

 System → Display → Value 4 display

Description

Select the measured value that is shown fourth on the local display.
Additional information:
The applicable unit of measure is specified in the "System units" submenu.

Selection• None• Volume flow
• Mass flow
• Conductivity *• Corrected conductivity *• Temperature *• Totalizer 1• Totalizer 2• Totalizer 3

Display damping

Navigation

 System → Display → Display damping

Description

Enter time constant (PT1 element) to set reaction time of the display to fluctuations in themeasured value.
Additional information:- The smaller the time constant the faster the display reacts to fluctuations in themeasured value.
- If the time constant is set to 0, damping is deactivated.

*

148

Visibility depends on order options or device settings

Endress+Hauser

Proline Promag 10 HART

User entry"System" menu

0.0 to 999.9 s

Rotation display

Navigation

 System → Display → Rotation display

Description

Select rotation angle of the display text to optimize local display readability.

Selection• Auto• 0 degree• 90 degree• 180 degree• 270 degree

Brightness
Navigation

 System → Display → Brightness

Description

Adjust brightness.

User entry

0 to 100 %Color scheme

Navigation

 System → Display → Color scheme

Description

Select preferred color scheme.

Selection• Light• DarkEndress+Hauser

149

"System" menu

Proline Promag 10 HART

5.7

"Software configuration" submenu

Navigation

 System → Software config.

‣ Software configuration
Activate SW option→  150

Software option overview→  150

Activate SW option



Navigation

 System → Software config. → Activate SW opt.

Description

Enter application package code or code of the functionality ordered separately to activate
it.
Additional information:- If a measuring device was ordered with an add-on software option, the activation code isprogrammed into the measuring device ex factory.
- After entering the activation code: Check whether the new software option is displayedin the "Software option overview" parameter and therefore active.
NOTE
If an an invalid code is entered the software options that have already been activated areinvalidated!
Before entering a new activation code: Create a record of the existing activation code.

User entry

Positive integer

Software option overview
Navigation

 System → Software config. → SW option overv.

Description

Displays all software options included in the order ex factory or ordered at a later date thathave been enabled via the operating interface.
Additional information:
If a new software option is not displayed after entering the activation code, the code
entered was inaccurate or invalid. In this case, contact the appropriate Endress+Hauser
sales organization to activate the software option.

User interface

150• Heartbeat Verification• Heartbeat Monitoring

Endress+Hauser

Proline Promag 10 HART

Index

Index
A

Activate SW option (Parameter) . . . . . . . . . . . . . . . 150
Active diagnostics (Submenu) . . . . . . . . . . . . . . . . . . 43
Actual diagnostics (Parameter) . . . . . . . . . . . . . . . . . 43
Alarm delay (Parameter) . . . . . . . . . . . . . . . . . . . . . . 54
Application (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 71
Assign behavior of diagnostic no. 043 (Parameter) . . . 55
Assign behavior of diagnostic no. 230 (Parameter) . . . 56
Assign behavior of diagnostic no. 231 (Parameter) . . . 57
Assign behavior of diagnostic no. 302 (Parameter) . . . 57
Assign behavior of diagnostic no. 376 (Parameter) . . . 58
Assign behavior of diagnostic no. 377 (Parameter) . . . 59
Assign behavior of diagnostic no. 441 (Parameter) . . . 60
Assign behavior of diagnostic no. 442 (Parameter) . . . 61
Assign behavior of diagnostic no. 443 (Parameter) . . . 61
Assign behavior of diagnostic no. 832 (Parameter) . . . 63
Assign behavior of diagnostic no. 833 (Parameter) . . . 64
Assign behavior of diagnostic no. 834 (Parameter) . . . 65
Assign behavior of diagnostic no. 835 (Parameter) . . . 66
Assign behavior of diagnostic no. 842 (Parameter) . . . 66
Assign behavior of diagnostic no. 937 (Parameter) . . . 67
Assign behavior of diagnostic no. 938 (Parameter) . . . 68
Assign behavior of diagnostic no. 961 (Parameter) . . . 69
Assign behavior of diagnostic no. 962 (Parameter) . . . 69
Assign diagnostic behavior (Parameter) . . . . . . . 33, 119
Assign flow direction check (Parameter) . . . . . . . . . 122
Assign frequency output (Parameter) . . . . . . . . . 28, 114
Assign limit (Parameter) . . . . . . . . . . . . . . . . . . 34, 119
Assign process variable (Parameter) . . . . . . . . . . . 14, 81
Assign pulse output 1 (Parameter) . . . . . . . . . . . 31, 110
Assign PV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 129
Assign QV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 131
Assign simulation process variable (Parameter) . . . . . 48
Assign status (Parameter) . . . . . . . . . . . . . . . . . 36, 123
Assign SV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 130
Assign TV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 130

Configuration (Submenu) . . . . . . . . . . . . . . . . . . . . . 59
Configuration counter (Parameter) . . . . . . . . . . . . . 134
Connectivity (Submenu) . . . . . . . . . . . . . . . . . . . . . 138
Control Totalizer 1 to n (Parameter) . . . . . . . . . . . . . . 82
Corrected conductivity (Parameter) . . . . . . . . . . . . . . 73
Current output 1 (Submenu) . . . . . . . . . . . . . . . . . . . 96
Current output 1 simulation (Parameter) . . . . . . . . . . 49
Current output value (Parameter) . . . . . . . . . . . . . . . 49
Current range output (Parameter) . . . . . . . . . . . 21, 101DBluetooth (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 138
Bluetooth configuration (Submenu) . . . . . . . . . . . . . 138
Brightness (Parameter) . . . . . . . . . . . . . . . . . . . . . . 149

Damping current output (Parameter) . . . . . . . . . 24, 104
Damping output 1 (Parameter) . . . . . . . . . . . . . . . . 117
Date / Time (Submenu) . . . . . . . . . . . . . . . . . . . . . . 139
Define access code (Wizard) . . . . . . . . . . . . . . . . . . 137
Density unit (Parameter) . . . . . . . . . . . . . . . . . . . 13, 78
Device (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 141
Device alarm simulation (Parameter) . . . . . . . . . . . . . 52
Device ID (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 127
Device management (Submenu) . . . . . . . . . . . . . . . . 133
Device name (Parameter) . . . . . . . . . . . . . . . . . . . 9, 141
Device reset (Parameter) . . . . . . . . . . . . . . . . . . . . . 134
Device revision (Parameter) . . . . . . . . . . . . . . . . . . . 126
Device tag (Parameter) . . . . . . . . . . . . . 9, 125, 133, 142
Device type (Parameter) . . . . . . . . . . . . . . . . . . . . . 127
Diagnostic configuration (Submenu) . . . . . . . . . . . . . 54
Diagnostic event category (Parameter) . . . . . . . . . . . . 52
Diagnostic event simulation (Parameter) . . . . . . . . . . 52
Diagnostic list (Submenu) . . . . . . . . . . . . . . . . . . . . . 45
Diagnostic settings (Submenu) . . . . . . . . . . . . . . . . . . 54
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 42
Diagnostics 1 (Parameter) . . . . . . . . . . . . . . . . . . . . . 45
Diagnostics 2 (Parameter) . . . . . . . . . . . . . . . . . . . . . 46
Diagnostics 3 (Parameter) . . . . . . . . . . . . . . . . . . . . . 46
Diagnostics 4 (Parameter) . . . . . . . . . . . . . . . . . . . . . 46
Diagnostics 5 (Parameter) . . . . . . . . . . . . . . . . . . . . . 47
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 146
Display damping (Parameter) . . . . . . . . . . . . . . . 39, 148
Display module (Submenu) . . . . . . . . . . . . . . . . . . . 144
Document
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

C

E

B

Calibration (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 92
Calibration factor (Parameter) . . . . . . . . . . . . . . . . . . 93
Color scheme (Parameter) . . . . . . . . . . . . . . . . . . . . 149
Commissioning (Wizard). . 6, 9, 10, 14, 16, 17, 18, 20, 21, 26, 28, 31, 32, 37, 39
Communication (Submenu) . . . . . . . . . . . . . . . . . . . 125
Communication established (Parameter) . . . . . . . . . 138
Conductivity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 72
Conductivity calibration factor (Parameter) . . . . . . . . 93
Conductivity damping time (Parameter) . . . . . . . . . . . 87
Conductivity measurement (Parameter) . . . . . . . . . . . 86
Conductivity temperature coefficient (Parameter) . . . . 87
Conductivity unit (Parameter) . . . . . . . . . . . . . . . 13, 78
Endress+Hauser

ECC duration (Parameter) . . . . . . . . . . . . . . . . . . . . . 94
ECC interval (Parameter) . . . . . . . . . . . . . . . . . . . . . . 95
ECC polarity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 95
ECC recovery time (Parameter) . . . . . . . . . . . . . . . . . 94
Electrode cleaning cycle (Parameter) . . . . . . . . . . . . . 94
Electrode cleaning cycle (Submenu) . . . . . . . . . . . . . . 94
Electronics (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 56
Empty pipe adjust value (Parameter) . . . . . . . . . . . . . 20
Empty pipe detection (Parameter) . . . . . . . . . . . . . . . 20
ENP version (Parameter) . . . . . . . . . . . . . . . . . . . . . 143
Enter access code (Parameter) . . . . . . . . . . . . . . . . . 136
Event category 043 (Parameter) . . . . . . . . . . . . . . . . 55
Event category 230 (Parameter) . . . . . . . . . . . . . . . . 56
151

IndexEvent category 231 (Parameter) . . . . . . . . . . . . . . . . 57
Event category 302 (Parameter) . . . . . . . . . . . . . . . . 57
Event category 376 (Parameter) . . . . . . . . . . . . . . . . 58
Event category 377 (Parameter) . . . . . . . . . . . . . . . . 59
Event category 441 (Parameter) . . . . . . . . . . . . . . . . 60
Event category 442 (Parameter) . . . . . . . . . . . . . . . . 61
Event category 443 (Parameter) . . . . . . . . . . . . . . . . 62
Event category 832 (Parameter) . . . . . . . . . . . . . . . . 64
Event category 833 (Parameter) . . . . . . . . . . . . . . . . 64
Event category 834 (Parameter) . . . . . . . . . . . . . . . . 65
Event category 835 (Parameter) . . . . . . . . . . . . . . . . 66
Event category 842 (Parameter) . . . . . . . . . . . . . . . . 67
Event category 937 (Parameter) . . . . . . . . . . . . . . . . 68
Event category 938 (Parameter) . . . . . . . . . . . . . . . . 68
Event category 961 (Parameter) . . . . . . . . . . . . . . . . 69
Event category 962 (Parameter) . . . . . . . . . . . . . . . . 70
Extended order code 1 (Parameter) . . . . . . . . . . . . . 143
Extended order code 2 (Parameter) . . . . . . . . . . . . . 143
Extended order code 3 (Parameter) . . . . . . . . . . . . . 143

F

Failure behavior current output (Parameter) . . . . 24, 105
Failure current (Parameter) . . . . . . . . . . . . . . . . 25, 105
Failure frequency (Parameter) . . . . . . . . . . . . . . 30, 117
Failure mode (Parameter) 16, 30, 37, 83, 112, 117, 123
Firmware version (Parameter) . . . . . . . 9, 142, 144, 145
Fixed current (Parameter) . . . . . . . . . . . . . . . . . 24, 102
Fixed density (Parameter) . . . . . . . . . . . . . . . . . . . . . 88
Flow damping (Parameter) . . . . . . . . . . . . . . . . . 16, 86
Flow override (Parameter) . . . . . . . . . . . . . . . . . . . . . 86
Frequency output 1 simulation (Parameter) . . . . . . . . 50
Frequency output 1 value (Parameter) . . . . . . . . . . . . 50
Full pipe adjust value (Parameter) . . . . . . . . . . . . . . . 20

G

Guidance (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

H

Hardware revision (Parameter) . . . . . . . . . . . . . . . . 128
HART address (Parameter) . . . . . . . . . . . . . . . . . . . 126
HART configuration (Submenu) . . . . . . . . . . . . . . . . 125
HART date code (Parameter) . . . . . . . . . . . . . . . . . . 128
HART descriptor (Parameter) . . . . . . . . . . . . . . . . . . 127
HART message (Parameter) . . . . . . . . . . . . . . . . . . . 128
HART revision (Parameter) . . . . . . . . . . . . . . . . . . . 127
HART short tag (Parameter) . . . . . . . . . . . . . . . . . . 125
Heartbeat Technology (Submenu) . . . . . . . . . . . . . . . 53

I

Information (Submenu) . . . . . . . . . . . . . . . . . . 126, 141
Installation direction (Parameter) . . . . . . . . . . . . . . . 91
Integration time (Parameter) . . . . . . . . . . . . . . . . . . . 92
Invert output signal (Parameter) . . . . . . . . . . . . . . . 124

L

Language (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 146
Locking status (Parameter) . . . . . . . . . . . . . . . . . . . 133
Low flow cut off (Parameter) . . . . . . . . . . . . . . . . 18, 88
Low flow cut off (Submenu) . . . . . . . . . . . . . . . . . . . . 88
Lower range value output (Parameter) . . . . . . . . 22, 102
152

Proline Promag 10 HARTM

Manufacturer (Parameter) . . . . . . . . . . . . . . . . . . . . 144
Manufacturer ID (Parameter) . . . . . . . . . . . . . . . . . 127
Mass flow (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 72
Mass flow unit (Parameter) . . . . . . . . . . . . . . . . . 12, 77
Mass unit (Parameter) . . . . . . . . . . . . . . . . . . . . . 12, 77
Maximum frequency value (Parameter) . . . . . . . 29, 116
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 71
Measuring mode (Parameter) . . . . . . . . . . . . . 111, 114
Measuring mode current output (Parameter) . . . . . . . 97
Measuring period (Parameter) . . . . . . . . . . . . . . . . . . 92
Measuring value at maximum frequency (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29, 116
Measuring value at minimum frequency (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29, 116
Menu
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 71
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 42
Guidance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 132
Minimum frequency value (Parameter) . . . . . . . 28, 115

N

Nominal diameter (Parameter) . . . . . . . . . . . . . . . . . 92

O

Off value low flow cutoff (Parameter) . . . . . . . . . . 19, 89
On value low flow cutoff (Parameter) . . . . . . . . . . 19, 89
Operating mode (Parameter) . . . . . . . . . . . . . . . 26, 108
Operating time (Parameter) . . . . . . . . . . . . . . . . . . . . 44
Operating time from restart (Parameter) . . . . . . . . . . 44
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . 142
Output (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 128
Output current 1 (Parameter) . . . . . . . . . . . . . . . . . 106
Output frequency 1 (Parameter) . . . . . . . . . . . . . . . 118

P

Partially filled pipe detection (Submenu) . . . . . . . . . . 89
Preset value 1 to n (Parameter) . . . . . . . . . . . . . . . . . 83
Pressure shock suppression (Parameter) . . . . . . . . . . 17
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . . 43
Primary variable (PV) (Parameter) . . . . . . . . . . . . . . 129
Process (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Process parameters (Submenu) . . . . . . . . . . . . . . . . . 85
Process variable current output (Parameter) . . . . . 21, 96
Process variable value (Parameter) . . . . . . . . . . . . . . . 49
Properties (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 54
Pulse output 1 (Parameter) . . . . . . . . . . . . . . . . . . . 113
Pulse output simulation 1 (Parameter) . . . . . . . . . . . . 50
Pulse value 1 (Parameter) . . . . . . . . . . . . . . . . . . . . . 51
Pulse width (Parameter) . . . . . . . . . . . . . . . . . . 31, 112
Pulse/frequency/switch output 1 (Submenu) . . . . . . 107

Q

Quaternary variable (QV) (Parameter) . . . . . . . . . . . 131

R

Reset all totalizers (Parameter) . . . . . . . . . . . . . . . . . 80
Reset Maintenance code (Parameter) . . . . . . . . . . . . 136

Endress+Hauser

Proline Promag 10 HART

Rotation display (Parameter) . . . . . . . . . . . . . . . . . . 149

S

Secondary variable (SV) (Parameter) . . . . . . . . . . . . 130
Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 55, 85
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . . . . 91
Sensor electronic module (ISEM) (Submenu) . . . . . . 144
Serial number (Parameter) . . . . . . . . . . . . . . . . . . 9, 142
Set date/time (Parameter) . . . . . . . . . . . . . . . . . . . . . 41
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 48
Software configuration (Submenu) . . . . . . . . . . . . . . 150
Software option overview (Parameter) . . . . . . . . . . . 150
Software revision (Parameter) . . . . . . . . . . . . . . . . . 128
Submenu
Active diagnostics . . . . . . . . . . . . . . . . . . . . . . . . 43
Bluetooth configuration . . . . . . . . . . . . . . . . . . . 138
Calibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 92
Communication . . . . . . . . . . . . . . . . . . . . . . . . . 125
Configuration . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Connectivity . . . . . . . . . . . . . . . . . . . . . . . . . . . . 138
Current output 1 . . . . . . . . . . . . . . . . . . . . . . . . . 96
Date / Time . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 141
Device management . . . . . . . . . . . . . . . . . . . . . . 133
Diagnostic configuration . . . . . . . . . . . . . . . . . . . . 54
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Diagnostic settings . . . . . . . . . . . . . . . . . . . . . . . . 54
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 146
Display module . . . . . . . . . . . . . . . . . . . . . . . . . . 144
Electrode cleaning cycle . . . . . . . . . . . . . . . . . . . . 94
Electronics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
HART configuration . . . . . . . . . . . . . . . . . . . . . . 125
Heartbeat Technology . . . . . . . . . . . . . . . . . . . . . 53
Information . . . . . . . . . . . . . . . . . . . . . . . . 126, 141
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . 88
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 71
Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 128
Partially filled pipe detection . . . . . . . . . . . . . . . . . 89
Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Process parameters . . . . . . . . . . . . . . . . . . . . . . . 85
Properties . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Pulse/frequency/switch output 1 . . . . . . . . . . . . 107
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55, 85
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 91
Sensor electronic module (ISEM) . . . . . . . . . . . . . 144
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Software configuration . . . . . . . . . . . . . . . . . . . . 150
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
Totalizer handling . . . . . . . . . . . . . . . . . . . . . . . . 80
Totalizers . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
User management . . . . . . . . . . . . . . . . . . . . . . . 135
Switch output function (Parameter) . . . . . . . . . . 32, 118
Switch output simulation 1 (Parameter) . . . . . . . . . . . 51
Switch state 1 (Parameter) . . . . . . . . . . . . . . . . 51, 124
Switch-off delay (Parameter) . . . . . . . . . . . . . . . 36, 122
Switch-off value (Parameter) . . . . . . . . . . . . . . . 36, 122
Switch-on delay (Parameter) . . . . . . . . . . . . . . . 36, 122
Endress+Hauser

Index

Switch-on value (Parameter) . . . . . . . . . . . . . . . 35, 121
System (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 132
System units (Submenu) . . . . . . . . . . . . . . . . . . . . . . 75

T

Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Temperature (Parameter) . . . . . . . . . . . . . . . . . . . . . 73
Temperature damping time (Parameter) . . . . . . . . . . 88
Temperature unit (Parameter) . . . . . . . . . . . . . . . 13, 78
Tertiary variable (TV) (Parameter) . . . . . . . . . . . . . . 131
Time format (Parameter) . . . . . . . . . . . . . . . . . . . . . . 39
Time zone (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 40
Timestamp (Parameter) . . . . . . . . . . . 43, 44, 45, 46, 47
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 73
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . . . . 80
Totalizer handling (Submenu) . . . . . . . . . . . . . . . . . . 80
Totalizer operation mode (Parameter) . . . . . . . . . 15, 82
Totalizer overflow 1 to n (Parameter) . . . . . . . . . . . . . 74
Totalizer value 1 to n (Parameter) . . . . . . . . . . . . . . . 74
Totalizers (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 80

U

Unit totalizer 1 to n (Parameter) . . . . . . . . . . . . . 14, 81
Upper range value output (Parameter) . . . . . . . . 23, 104
User management (Submenu) . . . . . . . . . . . . . . . . . 135
User role (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 135

V

Value 1 display (Parameter) . . . . . . . . . . . . . . . . 37, 147
Value 2 display (Parameter) . . . . . . . . . . . . . . . . 38, 147
Value 3 display (Parameter) . . . . . . . . . . . . . . . . 38, 148
Value 4 display (Parameter) . . . . . . . . . . . . . . . . 39, 148
Value per pulse (Parameter) . . . . . . . . . . . . . . . 32, 111
Volume flow (Parameter) . . . . . . . . . . . . . . . . . . . . . 72
Volume flow unit (Parameter) . . . . . . . . . . . . . . . 10, 75
Volume unit (Parameter) . . . . . . . . . . . . . . . . . . . 11, 76

W

WizardCommissioning
6, 9, 10, 14, 16, 17, 18, 20, 21, 26, 28, 31, 32,
37,
39
Define access code . . . . . . . . . . . . . . . . . . . . . . . 137

Z

Zero point (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 93

153

*71575135*
71575135

www.addresses.endress.com