4.47
113.4

4.0
101.01SERIAL NUMBERLABELAC INPUT

2X
1.06
26.8

.81
20.5
1.00
25.4

5.43
138

765

DETAIL B
SCALE 1.5 : 1FUSE

2X.78±.0319.9±0.85.250±.005
133.35±0.13

ON/OFFSWITCHNEEDLE VALVEFLOW METERDISPLAY WITHMAGNETIC ACTUATEDKEYPAD18.31±.06
465±1.5

18.86±.05
479±1.3

B

4X.280±.005
7.11±0.13

6.968±.005
176.99±0.13

MAGNETIC STYLUS

FULL RADIUS

2X

2 PLACES

31/32".XX [.X].XXX [.XX].02 [.1].010 [.04]

UNLESS OTHERWISE SPECIFIEDDIMENSIONS ARE IN INCHES & [mm]TOLERANCES ARE
FRACTIONSDECIMALSANGLESAREVJPSDRAWN~ECO2REVISIONSDESCRIPTIONS

6.82
173.2

Panametrics LLC
1100 Technology Park Dr.
Billerica, MA 01821 USA

ORIGINATED FOR REV CONTROL

DWN
JPS
12/12/2013

A

1CKD
MG
12/12/13

APVD

REVJG
12/12/13

CD4PROPRIETARY INFORMATION- THIS DRAWINGCONTAINS PROPRIETARY INFORMATION OF
PANAMETRICS AND MAY NOT BE
VIEWED OR DISCLOSED TO OTHERS, EXCEPTWITH THE WRITTEN PERMISSION OF GENERALELECTRIC CO.

COPYRIGHT 2013
Panametrics

CSURFACE FINISH

2AURORA RACK MOUNT 19

MODEL NO.

N/A

SOLIDWORKS

GENERATED USING

QC

N/A

N/A

07705

N/A

08/21/13 SALES
N/ARN

N/A

CERT

MG

08/21/13 APP'DJG
CHECKED

08/21/13

THIRD ANGLE PROJECTION

1

N/A

D
SCALE

SIZE

TITLE

NTS

712-1890
DO NOT SCALE DWGDRAWING NUMBER

1A
SHEET 1 OF 1

OUTLINE & INSTALLATION,AURORA, RACK MOUNTADETAIL A
SCALE 2 : 14.67
118.7.87
22.1

19.92
506

2X.55
143A

8A

B

SAMPLE OUTLET

2X
1.30
33.1

2.87
73417.00
431.8

B2 X .31
7.9
1.61
40.9

1.2632

SAMPLE INLET

17.50
444.5

5

1. WEIGHT: 58 lbs. [26.31Kilograms]
2. RACK MOUNT COMPLIANT 4U
3. COLOR: METALLIC

2X

6BC

D7C8

NOTES:

Panametrics
Mounting and Wiring the Aurora
Rack Mount Configuration

916-140, Rev B
September 2022

Mounting the Aurora Rack Mount Configuration

IMPORTANT: The Aurora rack mount configuration is not suitable for use in hazardous areas.

Figure 1: Aurora Drawing for Rack Mounting (Dwg. 712-1890, rev. A)

1100 Technology Park Drive
Billerica, MA 01821-4111
Web: www.panametrics.com

Mounting the Aurora Rack Mount Configuration1.

Pre-wire the output connectors in accordance with wiring drawing.

2.

Use approved AC input power cord provided.

3.

Sample inlet/outlet connections to be sure they are leak free.

4.

For additional support use a Rack Rail Kit as shown below or equivalent.

Figure 2: Standard Rack Mount Rail Kit
5.

Install 4 fasteners (12-24 or 10-32 threaded fasteners) to retain the rack mount.

916-140B

2

Mounting the Aurora Rack Mount Configuration

Wiring the Aurora Rack Mount Configuration

Figure 3: Wiring Diagram
TB1 contains connections for the alarm relay and 4-20mA analog output ‘C’ and the Alarm relay contacts. NC isNormally closed contact, C is Common, and NO is normally open contact.
Use shielded 18 – 22 AWG (0.82-0.33 mm2) twisted pair wire for all I/O connections.
TB2 contains PORT1 and PORT 2 designated SCADA and Service respectively. Either port may be configured for RS232
or RS485.
To ensure compliance to EN requirements, a shielded cable is required for these connections and the shield must beterminated to the SHD terminal. Also, a ferrite cable clamp will need to be added close to the I/O connector. Theferrite cable clamp should have the characteristics shown in Table 1 below:Table 1: Typical Cable Clamp Impedance ()10 MHz77916-140B

25 MHz+125

100 MHz+210

250 MHz260

3

Mounting the Aurora Rack Mount Configuration

A typical cable clamp is shown in Figure 4 below.

Figure 4: Cable Clamp
For operation on RS-485, 2-wire, half-duplex bus, attach the RS485(+) to (+) and the RS485(-) to (-). If available,
make a third connection to the RS-485 network common to RTN. Connect shield wire from the cable to SHD terminal.
For operation on RS-232, connect RS232 (TXD) to (+), RS232(RXD) to (-), and RS-232 (GND) to RTN. Connect cableshield wire to SHD terminal.
For connection to a PC interface with Auroraview software, you may used the supplied 704-668-12 cable (RS-232)
w/SUB-D-9 connector to the tinned leads. (There is also a detachable AC power cord on the back panel.).
Wire the cable as follows in Table 2 below.

Color code

Note:Table 2: Wiring Table
Aurora Terminal(Port 1 Pin) (Port 2 Pin)White

TX+ (10) (14)Red

RX- (11) (15)GreenReturn

RTN (12) (16)Black

ShieldSHD (13) (13)The default configuration is as shipped:
Table 3: Default Configuration
BAUD Rate:
115,200

916-140B

Parity:Even

ID Note:1 for Port 1, 2 for Port 2

4

Mounting the Aurora Rack Mount Configuration

Figure 5: Back Panel Labels

Figure 6: Location of Model and Serial Number

916-140B

5

Mounting the Aurora Rack Mount Configuration

Figure 7: Power, CE and Inlet Labels

Figure 8: Back Panel Label - Wiring Diagram

916-140B

6Mounting the Aurora Rack Mount Configuration

Figure 9: Laser Safety Label

916-140B7