DIGITAL MULTIMETERS

KEW MATE SERIESUP to 120A AC/DC Multimeters
Unique Open Jaw Technology for AC/DC current measurements

KEW MATE 2012R

KEW MATE 2000

KEW MATE 2001

Innovative Multimeteres with current measurements up to 120A
The open jaws are thin, perfect to clamp wires even in tight spac
Kewmates are very handy and as reliable as traditional full size

φ10mmm
φ12m

2001 JAW
ActualSize

φ6mm

2000 JAW● TRMS (True Root Mean Square) (Only for 2012R)

Fig. A

Fig. B

Sinusoidal

Distorted

For sinusoidal signals, TRMS and Averaging instruments read the same value, see figure A.
For distorted signals, only the TRMS instrument
reads the correct value, see figure B.

TRMS reading

Average reading

Kewmate 2012R with TRMS readings is recommended for measuring distorted currents and voltages generated by PCs, energy saving lighting,inverters, several home appliances and generally all
devices powered by switching power supplies.

TRMS reading● KewMates 2000/2001/2012R come with Extra Features!!!!

• They can measure AC/DC currents up to: 60A, with Max. Resolution 0.1A AC/DC (for 2000)
100A, with Max. Resolution 0.1A AC/DC (for 2001)
120A, with Max. Resolution 0.01A AC/DC (for 2012R)• Diode/Capacitance measurements（Only for 2012R）• All equipped with a shock absorbing holster that incorporates test leads and open clamp sensor.

Average reading

A AC/DC.
ces.
e Multimeters.

The open jaws of Kewmates are perfect in tight spaces wherea normal clamp meter is too big or wires are too short.

Insulated and removable caps prevent shortening of IC legs for proving
high density component boards.

One test probe can be fixed to themeter to help get the job donesafer and faster.

Quick one touch DC zero adjustment for 2012R (rotary knob for2000 and 2001).

● Applications• Everyday multimeters for technicians, the Kewmates are designed for electrical and electronic testing in field service
and all applications where extra features, such as AC/DC current measurements up to 60/100/120A, simplifies working.
• Ideal for electrical and electronic troubleshooting in residential, commercial and industrial installations.
• Thanks to the open clamp sensor, Kewmates are perfect for reading currents in tight spaces where a normal clamp
meter would be unsafe or too big to use such as sub-distribution boards, consumer units, final circuits, connection
boxes, control panels or simply where the wires are too short.
• Optimized for measuring the output of a solar panel in photovoltaic installations.
• DC current measurements permit electrical testing and servicing of cars, electric forklifts, small electric cars or trolleys for cleaning, DC motor adjustable speed controllers, golf cars and trolleys, small electric cars and chairlifts forthe disabled, work platforms, UPS s, electric scooters and bikes, etc.
• Accurate True RMS readings of AC current and voltage also in presence of harmonics created by PC s, Inverter,
Electronic Ballast for florescent lamps, UPS s, etc. (only 2012R)

Measuring current in a Switchboard

Servicing a Forklift

Servicing a car

●2000/2001/2012R Specifications
2000

φ62001

φ10MAX
AC/DC 60A

AC V

3.400/34.00/340.0/600V（Input impedance: approx.10MΩ）±1.5%rdg±5dgt（50〜400Hz）

6.000/60.00/600.0V（Input impedance: approx.10MΩ）±1.5%rdg±5dgt（45〜400Hz）

100.0A

60.0A

AC A

±2%rdg±5dgt（50/60Hz）±2%rdg±5dgt

60.0A

100.0A±2%rdg±5dgt（50/60Hz）

Buzzer sounds below 30±10Ω

Buzzer sounds below 35±25Ω

Capacitance−2V ±3.0%rdg±5dgt Open-loop voltage: approx. 2.7V
400.0nF/4.000/40.00μF±2.5%rdg±10dgt（AC A）3.400/10.00kHz±0.1%rdg±1dgt（AC V）3.400/34.00/300.0kHz±0.1%rdg±1dgt

Frequency（Input sensitivity（Input sensitivity

Current: more than 15A
Voltage: more than 30V）

Current: more than 25A
Voltage: more than 30V）

φ6mm max.

φ10mm max.

Withstand voltage 3700V AC for 1 minute

Power sourceDimensions
WeightAccessories
Optional

60.00/120.0A±2.0%rdg±5dgt（45〜65Hz）

600.0Ω/6.000/60.00/600.0kΩ/6.000/60.00MΩ
±1.0%rdg±5dgt（600Ω/6/60/600kΩ）±2.0%rdg±5dgt（6MΩ）±3.0%rdg±5dgt（60MΩ）−

Applicablestandards

60.00/120.0A±2.0%rdg±8dgt（60A）±2.0%rdg±5dgt（120A）

340.0Ω/3.400/34.00/340.0kΩ/3.400/34.00MΩ
±1%rdg±3dgt（0~340kΩ）±5%rdg±5dgt（3.400MΩ）±15%rdg±5dgt（34.00MΩ）

Diode test

Conductor size

600.0mV/6.000/60.00/600.0V（Input impedance: approx. 10MΩ）±1.0%rdg±3dgt±2%rdg±5dgt

Continuity buzzer

MAXAC/DC 120A

（Input impedance: approx. 10MΩ）±1.5%rdg±4dgt

DC AΩ

φ12MAX
AC/DC 100A

340.0mV/3.400/34.00/340.0/600V

DC V

2012R

（AC A）100/1000Hz/10kHz±0.2%rdg±2dgt（100Hz）±0.1%rdg±1dgt（1000Hz/10kHz）（AC V）100/1000Hz/10/100/300.0kHz±0.2%rdg±2dgt（100Hz）±0.1%rdg±1dgt（1000Hz/10/100/300.0kHz）（Input sensitivity)

Current: more than 2A
Voltage: more than 2V［〜10kHz］/more than 20V［10k〜300kHz］
φ12mm max.

3540V AC for 5 seconds

IEC 61010-1 CAT.Ⅲ 300V, CAT.Ⅱ 600V Pollution degree 2
IEC 61010-031, IEC 61010-2-032, IEC 61326

IEC 61010-1 CAT.Ⅲ 300V, CAT.Ⅱ 600V Pollution degree 2
IEC 61010-031, IEC 61010-2-032, IEC 61326

×2R03（1.5V）*Continuous measuring time: approx. 45hours（Auto power save: approx. 10 minutes）

×2
R03（1.5V）*Continuous measuring time: DCV: approx. 150hours,ACA: approx. 25hours（Auto power save: approx. 15 minutes）

×87（W）
×24（D）
128（L）
mm×91（W）
×27（D）
128（L）
mm×92（W）×27（D）
128（L）mm

210g approx.
（including batteries）

220g approx.
（including batteries）

220g approx.（including batteries）

×2 Instruction manual
R03（1.5V）

9107（Carrying case［Soft］）

Please read the "Safety Warnings" in the instruction manual supplied with the instrument thoroughly and completelycorrect use. Failure to follow the safety rules can cause fire, trouble, electrical shock, etc. Therefore, make sure
Safety Warnings : forto operate the instrument on a correct power supply and voltage rating marked on each instrument.

For inquires or orders :ISO9001:2000
APPROVED BY BUREAU VERITAS Certification

In consideration of the environment, soy ink and recycled paper were used in this publication.

The contents of this leaflet are subject to change without notice.

KEW MATE 2000/2001/2012R-1E Jan. 10 AD