Hospital
respiratory care
V60 ventilator

Dedicated tosuccessful NIV
Because every breath matters and each one is diﬀerent,
patients need a ventilator that always follows their breathing
pattern, whatever their acuity. Philips Respironics V60
Ventilator combines Respironics ventilation expertisewith Philips focus on simplifying advanced health care.
The result is noninvasive ventilation excellence with aninvasive ventilation alternative and an interactive displaythat helps simplify patient management.

Key advantages:• ICU-grade NIV performance withenhanced monitoring and alarms• Treatment for a wide range of clinicalseverities and patients from pediatricto adult• Flexibility for intra-hospital transportwith an internal six-hour smart battery

The print quality of this copy is not an accurate representation of the original.

Highlights
The Philips Respironics V60 uses
auto-adaptive technology to help
ensure patient synchrony and therapy
acceptance. Improved signal processing
technology is finely tuned for both adultand pediatric patients.

Auto-adaptive triggering and cycling with Auto-Trak

Advanced NIV with Auto-Trak• Auto-adaptive leak compensation• Auto-adaptive inspiratory triggering• Auto-adaptive expiratory cycling
Hospital modes and options• AVAPS maintains a target tidal volume in a pressure‑limitedmode. It provides extra reassurance similar to a volumelimited mode with the advantages of a pressure‑limited mode.
• PCV can be used when full control of the patient’s breathing
pattern is required.
• CPAP with C-Flex option offers three levels of flow-based
expiratory pressure relief. This leads to enhanced sleep quality
and patient comfort, adding greater flexibility and improved
treatment acceptance.
• Proportional Pressure Ventilation (PPV) provides inspiratory
flow and pressure in proportion to the patient’s spontaneous
effort,1 thus improving patient control over their ventilation.2• Auto-Trak Plus is an option for a subset of patients who
may benefit from a customized titration of triggering andcycling criteria.

Intra-hospital transport with internal six-hour battery

The print quality of this copy is not an accurate representation of the original.

Patient types

Pediatric NIV

Adult
Pediatric (≥20kg)
Monitored parameters
Modes
Continuous positive airway pressure (CPAP)
Spontaneous with timed backup (S/T)
Pressure control ventilation (PCV)
Average volume assured pressure support (AVAPS)
Proportional pressure ventilation (PPV)* – optionalSettings

Breath phase/trigger indicator

Spont, timed, exhale

PIP

0 – 50cmH2O

Patient/total leak0 – 200 l/min btps

Patient trigger

0 – 100%

Respiratory rate0 – 90bpm

Ti/Ttot

0 – 91%

Minute volume

0 – 99.0 l/min btps

Tidal volume

0 – 3,500ml btps

C-Flex

Off, 1 – 3CPAP

4 – 25cmH2OGeneralEPAP

4 – 25cmH2O

Oxygen inlet pressure range

IPAP

4 – 40cmH2O

276 – 600kPa(40 – 87psig)

I-time (inspiratory time)

0.30 – 3.00sec

WeightMax P (AVAPS maximum IPAP)

6 – 40cmH2O

11.7kg (25.7lb)
with optional battery
10.6kg (23.3lb)
without optional battery

Dimensions

33.7cm (13.3in) height
39.4cm (15.5in) width
42.9cm (16.5in) depth

Min P (AVAPS minimum IPAP)

5 – 30cmH2O

O2 (oxygen percent)21 – 100%Ramp time

Off, 5 – 45min

Rate (respiratory rate)4 – 60BPM

Rise (rise time)

1–5Triggering and cycling

Auto-adaptive (Auto-Trak)

Auto-Trak Plus*

OptionalTrigger*Normal, +1 – 7E-Cycle*-2, -1, Normal, +1 – +6

AVAPS target tidal volume

200 – 2,000ml

Electrical
AC voltage

100 – 240VAC

AC frequency

50/60HzAC power

300VA

Battery operating time

Six hours in normal
conditions* May not be available in all markets

The print quality of this copy is not an accurate representation of the original.

Description

Legacy part number

12NC part number

Philips Respironics V60 ventilator with AVAPS, C-Flex (US)1053617989805611761

Philips Respironics V60 ventilator with AVAPS and C-Flex, International
excluding China and Japan1053614

989805612101

Philips Respironics V60 ventilator with AVAPS, C-Flex and PPV, International
excluding China and Japan

1053613989805628251

Philips Respironics V60 configuration kit

country specificcountry specificCommunication
ports(RS232, serial,
remote alarm)

Navigation ring

Large color
touchscreen

Six-hourinternal smart
battery

Rugged cartwith multiple
accessory
attachment
capabilities

Optional O2
manifold(rear of cart)

Optional O2tank holders

1

Gay P., Hess D. and Hill N. Noninvasive proportional assist ventilation
for acute respiratory insufficiency comparison with pressure supportventilation. Am J Respir Crit Care Med. 2001; 164:1606-1611© 2016 Koninklijke Philips N.V. All rights reserved.
Specifications are subject to change without notice.

2Younes M. Proportional assist ventilation, a new approach to 		ventilatory support. American Review of Respiratory Disease. 		
1992; 145(1):114 -120

www.philips.com/V60
4522 991 22631 * OCT 2016

The print quality of this copy is not an accurate representation of the original.