SFD-753114-103-10SF-P1

Rev 1.1

Description:
Model SFD-753114-103-10SF-P1 is a W Band amplitudedetector that can be used for full or narrow band applications.
The detector is zero biased and intended for small signal
detection purposes. With a distinct circuitry design and careful
diode selection, the detector exhibits high sensitivity andextremely flat output characteristics across the full waveguide
operating bandwidth. The detector is designed to have a 10
MHz video bandwidth and a 1 MΩ video output impedance.
The minimum detectable signal level is approximately -50 dBm.

Features:•••

Applications:

Full Waveguide Band Operation
High Sensitivity Without Tuning
High Sensitivity Stability Over Broad
Temperature Range•••

Radar Systems
Communication Systems
Test instrumentations

Electrical Specifications:Parameter
Frequency
Sensitivity*
Sensitivity Flatness
RF Input PowerRF Power HandlingVideo Bandwidth**
Detection Speed, Rise Time (50 Ohm Load)
Output Voltage Polarity
Specification Temperature
Operating Temperature

Minimum
75 GHzTypicalMaximum
110 GHz

SAGE Millimeter, Inc.

W-Band Amplitude Detector, Small Signal, Positive

800 mV/mW±2.0 dB-20 dBm+17 dBm
10 MHz
10 ns
Positive+25 ˚C-40 ˚C+85 ˚C*Note: The sensitivity is for the input signal level -20 dBm or below.
**Note: The video bandwidth can extend up to 10 GHz.

Mechanical Specifications:
Item
RF PortDC PortCase Material
FinishWeightOutlineSpecification
WR-10 Waveguide with UG-387/U-M Anti-Cocking Flange
SMA (F)
AluminumGold Plated
0.4 OzFD-W1-103-A
www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com

Copyright © 2022 by Eravant

Page | 1

SFD-753114-103-10SF-P1Rev 1.1

Typical Sensitivity vs. Frequency
Pin = -20 dBm

Sensitivity (mV/mW)

10000

1000100

101758085

9095100105110

Frequency (GHz)

SAGE Millimeter, Inc.

W-Band Amplitude Detector, Small Signal, Positive

Typical Detected Voltage vs. Input Power20075 GHz

Detected Voltage (mV)

92.5 GHz
110 GHz

150

100

50

0-25-20-15-10-5Input Power (dBm)

www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com
Copyright © 2022 by Eravant

Page | 2

SFD-753114-103-10SF-P1

Rev 1.1

Mechanical Outline: (Unless otherwise specified, all dimensions are in inches)

SAGE Millimeter, Inc.

W-Band Amplitude Detector, Small Signal, Positive

Note:
•••
•
•

All data presented is collected from a sample lot. Actual data may vary unit to unit.
All testing was performed under +25˚C case temperature.
The negative output voltage polarity is offered under the model number SFD-753114-103-10SFN1.
The amplitude detector is a small signal detector. The sensitivity shown is for RF signal -20 dBm
or lower.
Eravant reserves the right to change the information presented without notice.

Caution:••••Exceeding absolute maximum ratings will damage the device.
The device is static sensitive. Always follow ESD rules when working with the device.
Any foreign objects in the waveguide will cause performance degradation and can possibly
damage the device.
Proper torque, 8.0 ± 0.15 inch-pounds (0.92 ± 0.05 Nm), should be applied. Eravant torquewrench, model SCH-08008-S1, is highly recommended.

www.eravant.com | 501 Amapola Avenue, Torrance, CA 90501
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: support@eravant.com
Copyright © 2022 by Eravant

Page | 3