High brightness:
Triple the power of conventional models**Compared with Sugawara's BS-series stroboscopes

Wide flashing range(30 - 36,000 FPM)

Single-handed operation
Lightweight at only 900 g

Battery-operated
Can be used while charging

Multiple modes of operation
Internal / external trigger and tachometer functions

Check the motion without stopping it.
Bright and sharp!
A one-hand operable,
general-purpose stroboscope
Applications

Stroboscopes are being applied in various fields such as,
measuring the motor speed, checking rapidly moving machinery parts,
surface inspection in steel and printing industries, and study andanalysis of high-speed phenomena in R&D.

ࠉFlawless Illumination
Short pulses for sharp stationary images
Uniform light by matte-finished mirror

At a distance of 1 m, the device uniformly illuminates anarea 60 cm in diameter.

OEL display easy to seeeven in bright places

Wide range ofeasy-to-use functions

60cm

100 cmSpecifications
Internal trigger: 30 - 36000 FPM
External trigger: 0 - 36000 FPM

Three levels ofbrightness control,
L, M, and H

Tachometer measurement range

30 - 120000 RPM

Output power

Switchable between 5 W / 10 W / 15 W

Flash duration

2.4 µs typical (full width at half maximum)

Maximum output energy

0.3 J/F

Operation modes

Internal trigger mode / External trigger mode / Tachometer mode

External trigger
input signals

Voltage signal, TTL signal, open collector, relay, contact signal

ࠉSharp stationary images made possible
due to a short pulse illumination
The shorter the flash duration, the sharper are thestationary images. The MIGHTY STROBE X-1's flash
duration is exceptionally short (several tens oftimes shorter than LED stroboscopes),allowing even
rapidly moving objects to be observed clearly.
The MIGHTY STROBE X-1's flash duration and intensity
compared to LED stroboscopes on the market

ࠉVoltage Signal: Low Level 0 - 3.6 V
ࠉHigh Level 5 - 26.4 V, Pulse width 10μsec min.
ࠉ
ࠉTTL Signal: Low Level 0 - 1 V
High Level 2 - 5 V, Pulse width 10μsec min.

External trigger output signals

Photo coupler output: 8 mA / DC 30 V

Flash lamp

High-brightness Xenon Flash Lamp FX-6SP

Battery

Lithium-ion battery (internal)

4000

Relative intensity㸦㸣㸧Flash range

X-1
2.4 μsec2000

Battery life: 4 h at 5 W; 1.5 h at 15 W typical
Recharge time: 2 h typicalAC adapter

AC 100 - 240 V, 50 / 60 Hz

Weight900 g

LED stroboscope

10000200

Flash duration 㸦μsec㸧

(Tested at flash frequency 3000 FPM and at a 300 mm distance.)

Dimensions (L×W×H)

182 × 122 × 294 mm

Standard accessories

AC Adaptor, Hand strap, Operation manual

Optional Accessories

LED stroboscopes have long flash durations
to achieve flash intensity.

● Sensors for external inputs

Input signal cable: 8CK-602-182
Output signal cable: 8CK-402-182Proximity / Photoelectric sensors for external input:ࠉࠉࠉࠉࠉࠉࠉࠉ (Please refer to our website.)

Proximity sensor

Photoelectric sensor

Read the instruction manual before using the device to ensure safety and to avoid damage to the device.

Individual trigger input & output jacks
allow daisy-chain of strobes andsupply power to the external sensors.

Tripod mountable

Photoelectric / Proximity sensors canbe used to measure rotational speed.

Head office/Tokyo sales office: 8-2 Minami-Kurokawa, Asao-ku,Kawasaki-shi, Kanagawa Prefecture 215-0034, JapanTel. 044-989-7320, fax 044-989-7338
Osaka sales office: 6-17 Yokomakura-Nishi,
Higashiosaka-shi, Osaka Prefecture 578-0956, Japan
Tel. 072-966-1061, fax 072-966-0961
Nagoya sales office: 1-2-29 Kamimaezu, Naka-ku,
Nagoya-shi, Aichi Prefecture 460-0013, Japan
Tel. 052-331-6562, fax 052-331-6604
URL: http://www.sugawara-labs.co.jp/
E-mail: info@sugawara-labs.co.jp●Specifications and performance are subject to change without prior notice.

20YC1603en