BA02261F/00/EN/01.23-00
71627296
2023-09-01

Products

Solutions

01.00.zz (Device firmware)

Operating Instructions
Micropilot FMR60B
PROFIBUS PA
Free space radar

Services

Micropilot FMR60B PROFIBUS PA

1.

Order code: XXXXX-XXXXXX
Ser. no.:XXXXXXXXXXXX
Ext. ord. cd.: XXX.XXXX.XX

Serial number

2. www.endress.com/deviceviewer

Endress+Hauser
Operations App

3.

A0023555• Make sure the document is stored in a safe place such that it is always available whenworking on or with the device• Avoid danger to individuals or the facility: read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures
The manufacturer reserves the right to modify technical data without prior notice. The
Endress+Hauser sales organization will supply you with current information and updates
to these instructions.

2

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 5

1.1
1.2

5.4

6

Electrical connection . . . . . . . . . . . . . . 22

1.4
1.5

Purpose of this document . . . . . . . . . . . . . . . . . 5
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5
1.2.1 Safety symbols . . . . . . . . . . . . . . . . . . 5
1.2.2 Symbols for certain types of
information and graphics . . . . . . . . . . . 5
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 6
1.3.1 Technical Information (TI) . . . . . . . . . 6
1.3.2 Brief Operating Instructions (KA) . . . . . 6
1.3.3 Safety Instructions (XA) . . . . . . . . . . . . 6
Terms and abbreviations . . . . . . . . . . . . . . . . . 7
Registered trademarks . . . . . . . . . . . . . . . . . . . 7

Integrated antenna, PEEK
40 mm (1.5 in) . . . . . . . . . . . . . . . . . 17
5.3.5 Turning the housing . . . . . . . . . . . . . 18
5.3.6 Turning the display module . . . . . . . . 18
5.3.7 Changing the installation position of
the display module . . . . . . . . . . . . . . 19
5.3.8 Closing the housing covers . . . . . . . . . 21
Post-mounting check . . . . . . . . . . . . . . . . . . . 21

6.1

2

Basic safety instructions . . . . . . . . . . . . 8

6.2

2.1
2.2
2.3
2.4
2.5
2.6
2.7

Requirements for personnel . . . . . . . . . . . . . . . 8
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . 9
Operational safety . . . . . . . . . . . . . . . . . . . . . . 9
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . 9
IT security . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Device-specific IT security . . . . . . . . . . . . . . . 10

6.3

3

Product description . . . . . . . . . . . . . . . . 11

6.4

Connecting requirements . . . . . . . . . . . . . . . 22
6.1.1 Cover with securing screw . . . . . . . . . 22
6.1.2 Potential equalization . . . . . . . . . . . . 22
Connecting the device . . . . . . . . . . . . . . . . . . 23
6.2.1 Supply voltage . . . . . . . . . . . . . . . . . 23
6.2.2 Cable specification . . . . . . . . . . . . . . . 24
6.2.3 Overvoltage protection . . . . . . . . . . . 24
6.2.4 Wiring . . . . . . . . . . . . . . . . . . . . . . . 25
6.2.5 Terminal assignment . . . . . . . . . . . . 26
6.2.6 Cable entries . . . . . . . . . . . . . . . . . . . 27
6.2.7 Available connectors . . . . . . . . . . . . 27
Ensuring the degree of protection . . . . . . . . . 28
6.3.1 Cable entries . . . . . . . . . . . . . . . . . . . 28
Post-connection check . . . . . . . . . . . . . . . . . . 28

3.1

Product design . . . . . . . . . . . . . . . . . . . . . . . . 11

7

Operation options . . . . . . . . . . . . . . . . . 29

4

Incoming acceptance and productidentification . . . . . . . . . . . . . . . . . . . . . 12

7.1
7.2

4.1
4.2

Incoming acceptance . . . . . . . . . . . . . . . . . . . 12
Product identification . . . . . . . . . . . . . . . . . . . 12
4.2.1 Nameplate . . . . . . . . . . . . . . . . . . . . 12
4.2.2 Manufacturer address . . . . . . . . . . . . 13
Storage and transport . . . . . . . . . . . . . . . . . . 13
4.3.1 Storage conditions . . . . . . . . . . . . . . . 13
4.3.2 Transporting the product to themeasuring point . . . . . . . . . . . . . . . . 13

Overview of operation options . . . . . . . . . . . . 29
Operating keys and DIP switches on the
PROFIBUS PA electronic insert . . . . . . . . . . . . 29
7.2.1 Hardware addressing . . . . . . . . . . . . . 29
7.2.2 Software addressing . . . . . . . . . . . . . 30
Structure and function of the operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
7.3.1 User roles and related accessauthorization . . . . . . . . . . . . . . . . . . 30
Access to operating menu via local display . . . 30
7.4.1 Device display (optional) . . . . . . . . . . 30
7.4.2 Operation via Bluetooth® wirelesstechnology (optional) . . . . . . . . . . . . 31
Access to the operating menu via theoperating tool . . . . . . . . . . . . . . . . . . . . . . . . 32
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . 32
7.6.1 Function scope . . . . . . . . . . . . . . . . . 32
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
7.7.1 Function range . . . . . . . . . . . . . . . . . 32

1.3

4.3

5

Mounting . . . . . . . . . . . . . . . . . . . . . . . . . 14

5.1
5.2

General instructions . . . . . . . . . . . . . . . . . . . . 14
Mounting requirements . . . . . . . . . . . . . . . . . 14
5.2.1 Internal vessel fittings . . . . . . . . . . . . 14
5.2.2 Avoiding interference echoes . . . . . . . 15
5.2.3 Vertical alignment of antenna axis . . . 15
5.2.4 Radial alignment of the antenna . . . . 15
5.2.5 Optimization options . . . . . . . . . . . . . 15
Mounting the device . . . . . . . . . . . . . . . . . . . 15
5.3.1 Encapsulated antenna, PVDF
40 mm (1.5 in) . . . . . . . . . . . . . . . . . 15
5.3.2 Drip-off antenna PTFE
50 mm (2 in) . . . . . . . . . . . . . . . . . . 16
5.3.3 Integrated antenna, PEEK
20 mm (0.75 in) . . . . . . . . . . . . . . . . 17

5.3

Endress+Hauser

5.3.4

7.3

7.4

7.5
7.6
7.7

8

System integration . . . . . . . . . . . . . . . . 34

8.1

Protocol-specific data . . . . . . . . . . . . . . . . . . 34
8.1.1 PROFIBUS PA . . . . . . . . . . . . . . . . . . 34
Device master file (GSD) . . . . . . . . . . . . . . . . . 35
8.2.1 Manufacturer-specific GSD . . . . . . . . 35
8.2.2 Profile GSD . . . . . . . . . . . . . . . . . . . . 35
8.2.3 Compatibility with other devices . . . . 36

8.2

3

Table of contents

8.3

Cyclic data transmission . . . . . . . . . . . . . . . . . 36
8.3.1 Block model . . . . . . . . . . . . . . . . . . . 36
8.3.2 Description of the modules . . . . . . . . 37

9Commissioning . . . . . . . . . . . . . . . . . . . . 40

9.1
9.2
9.3

9.6
9.7

Preliminaries . . . . . . . . . . . . . . . . . . . . . . . . 40
Function check . . . . . . . . . . . . . . . . . . . . . . . 40
Connecting via FieldCare and DeviceCare . . . . 40
9.3.1 Via PROFIBUS PA protocol . . . . . . . . . 40
9.3.2 Via service interface (CDI) . . . . . . . . . 41
Setting the operating language . . . . . . . . . . . 41
9.4.1 Local display . . . . . . . . . . . . . . . . . . . 41
9.4.2 Operating tool . . . . . . . . . . . . . . . . . . 41
Configuring the device . . . . . . . . . . . . . . . . . 42
9.5.1 Level measurement in liquids . . . . . . . 42
9.5.2 Commissioning with theCommissioning wizard . . . . . . . . . . . 42
Recording the Echo curve . . . . . . . . . . . . . . . . 43
"Simulation" submenu . . . . . . . . . . . . . . . . . . . 43

10Operation . . . . . . . . . . . . . . . . . . . . . . . . . 44

10.1
10.2
10.3
10.4

Reading off the device locking status . . . . . . . 44
Reading off measured values . . . . . . . . . . . . . 44
Adapting the device to process conditions . . . . 44
Heartbeat Technology (optional) . . . . . . . . . . 44
10.4.1 Heartbeat Verification . . . . . . . . . . . . 44
10.4.2 Heartbeat Verification/Monitoring . . 44

11

Diagnostics and troubleshooting . . . 45

9.4
9.5

11.1

General troubleshooting . . . . . . . . . . . . . . . . . 45
11.1.1 General errors . . . . . . . . . . . . . . . . . . 45
11.2 Error - SmartBlue operation . . . . . . . . . . . . . . 45
11.3 Diagnostic message . . . . . . . . . . . . . . . . . . . . 46
11.3.1 Diagnostic information on local
display . . . . . . . . . . . . . . . . . . . . . . . 47
11.3.2 Diagnostic information in theoperating tool . . . . . . . . . . . . . . . . . . 47
11.3.3 Status signal . . . . . . . . . . . . . . . . . . . 47
11.3.4 Diagnostic event and event text . . . . . 48
11.4 Remedy informationCalling up . . . . . . . . . . . 48
11.4.1 Graphic display with keys . . . . . . . . . 48
11.4.2 Operating menu . . . . . . . . . . . . . . . . 49
11.5 Adapting the diagnostic information . . . . . . . 49
11.6 List of diagnostic events . . . . . . . . . . . . . . . . . 49
11.7 Event logbook . . . . . . . . . . . . . . . . . . . . . . . . 51
11.7.1 Event history . . . . . . . . . . . . . . . . . . . 51
11.7.2 Filtering the event logbook . . . . . . . . 52
11.7.3 Overview of information events . . . . . 52
11.8 Resetting the device . . . . . . . . . . . . . . . . . . . . 53
11.8.1 Reset password via operating tool . . . 53
11.8.2 Reset device via operating tool . . . . . 53
11.8.3 Resetting the device via keys on theelectronic insert . . . . . . . . . . . . . . . . 53
11.9 Device information . . . . . . . . . . . . . . . . . . . . 54
11.10 Firmware history . . . . . . . . . . . . . . . . . . . . . . 54

4

Micropilot FMR60B PROFIBUS PA

12Maintenance . . . . . . . . . . . . . . . . . . . . . . 55

12.1
12.2

Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . 55
Seals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55

13

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56

13.1

General information . . . . . . . . . . . . . . . . . . . 56
13.1.1 Repair concept . . . . . . . . . . . . . . . . . 56
13.1.2 Repairs to Ex-approved devices . . . . . 56
13.2 Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . 56
13.3 Replacement . . . . . . . . . . . . . . . . . . . . . . . . . 56
13.3.1 HistoROM . . . . . . . . . . . . . . . . . . . . . 56
13.4 Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
13.5 Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57

14

Accessories . . . . . . . . . . . . . . . . . . . . . . . 58

14.1 Weather protection cover 316L . . . . . . . . . . . 58
14.2 Plastic weather protection cover . . . . . . . . . . . 58
14.3 Mounting bracket, adjustable . . . . . . . . . . . . . 59
14.3.1 Dimensions . . . . . . . . . . . . . . . . . . . . 59
14.3.2 Scope of delivery . . . . . . . . . . . . . . . . 60
14.4 Remote display FHX50B . . . . . . . . . . . . . . . . . 60
14.5 Gas-tight feedthrough . . . . . . . . . . . . . . . . . . 61
14.6 Field Xpert SMT70 . . . . . . . . . . . . . . . . . . . . . 61
14.7 DeviceCare SFE100 . . . . . . . . . . . . . . . . . . . . 61
14.8 FieldCare SFE500 . . . . . . . . . . . . . . . . . . . . . 61
14.9 RID14 . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
14.10 RID16 . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
14.11 Fieldgate SFG500 . . . . . . . . . . . . . . . . . . . . . 63

15

Technical data . . . . . . . . . . . . . . . . . . . . 64

15.1 Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
15.2 Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
15.3 Environment . . . . . . . . . . . . . . . . . . . . . . . . .
15.4 Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

64
70
72
85

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 88

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

About this document

1

About this document

1.1

Purpose of this document

These Operating Instructions contain all the information that is required in the various
phases of the life cycle of the device: from product identification, incoming acceptance andstorage, to mounting, connection, operation and commissioning through totroubleshooting, maintenance and disposal.

1.2

Symbols

1.2.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

1.2.2

Symbols for certain types of information and graphics

Permitted
Procedures, processes or actions that are permitted
Forbidden
Procedures, processes or actions that are forbidden
Tip
Indicates additional information
Reference to documentationA

Reference to page
Reference to graphic
Notice or individual step to be observed
1. , 2. , 3.
Series of steps

Result of a step
1, 2, 3, ...
Item numbers
A, B, C, ...
Views

Endress+Hauser

5About this document

Micropilot FMR60B PROFIBUS PA

1.3

Documentation

The following document types are available in the Downloads section of the Endress+Hauser website (www.endress.com/downloads):
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• Device Viewer (www.endress.com/deviceviewer): Enter the serial number from thenameplate• Endress+Hauser Operations app: Enter serial number from nameplate or scanmatrix code on nameplate.

1.3.1

Technical Information (TI)

Planning aid
The document contains all the technical data on the device and provides an overview of
the accessories and other products that can be ordered for the device.

1.3.2

Brief Operating Instructions (KA)

Guide that takes you quickly to the 1st measured value
The Brief Operating Instructions contain all the essential information from incoming
acceptance to initial commissioning.

1.3.3

Safety Instructions (XA)

Depending on the approval, the following Safety Instructions (XA) are supplied with thedevice. They are an integral part of the Operating Instructions.
The nameplate indicates the Safety Instructions (XA) that are relevant to the device.

6

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

1.4

About this document

Terms and abbreviations

MWP
Maximum working pressure
The MWP is indicated on the nameplate.
εr (Dk value)
Relative dielectric constantOperating tool
The term "operating tool" is used in place of the following operating software:
SmartBlue (app), for operation using an Android or iOS smartphone or tablet
PLC
Programmable logic controller (PLC)
CDI
Common Data Interface

1.5

Registered trademarks

PROFIBUS®
PROFIBUS and the associated trademarks (The Association Trademark, the Technology
Trademarks, the Certification Trademark and the Certified by PI Trademark) are registeredtrademarks of the PROFIBUS User Organization e.V. (Profibus User Organization),
Karlsruhe - Germany
Bluetooth®
The Bluetooth® word mark and logos are registered trademarks owned by the Bluetooth
SIG, Inc. and any use of such marks by Endress+Hauser is under license. Other trademarks
and trade names are those of their respective owners.
Apple®
Apple, the Apple logo, iPhone, and iPod touch are trademarks of Apple Inc., registered in
the U.S. and other countries. App Store is a service mark of Apple Inc.
Android®
Android, Google Play and the Google Play logo are trademarks of Google Inc.
KALREZ®, VITON®
Registered trademarks of DuPont Performance Elastomers L.L.C., Wilmington, DE USA

Endress+Hauser

7Basic safety instructions

Micropilot FMR60B PROFIBUS PA

2Basic safety instructions

2.1

Requirements for personnel

The personnel for installation, commissioning, diagnostics and maintenance must fulfill
the following requirements:‣ Trained, qualified specialists must have a relevant qualification for this specific function
and task.
‣ Personnel must be authorized by the plant owner/operator.
‣ Be familiar with federal/national regulations.
‣ Before starting work: personnel must read and understand the instructions in the
manual and supplementary documentation as well as the certificates (depending on
the application).
‣ Personnel must follow instructions and comply with general policies.
The operating personnel must fulfill the following requirements:‣ Personnel are instructed and authorized according to the requirements of the task by
the facility's owner-operator.

‣ Personnel follow the instructions in this manual.

2.2

Intended use

Application and media
The measuring device described in these Operating Instructions is intended for continuous,
non-contact level measurement in liquids, pastes and sludges. Due to its operating
frequency of approx. 80 GHz, a maximum radiated peak power of <1.5 mW and an
average output power of <70 µW, unrestricted use outside of closed, metallic vessels is
also permitted (for example over basins or open channels). Operation does not pose any
danger whatsoever to humans and animals.
If the limit values specified in the "Technical data" and the conditions listed in the
instructions and additional documentation are observed, the measuring device may be
used only for the following measurements:‣ Measured process variables: level, distance, signal strength‣ Calculated process variables: volume or mass in vessels of any shape; flow rate throughmeasuring weirs or channels (calculated based on the level using the linearization
functionality)
To ensure that the measuring device remains in proper condition for the operation time:‣ Use the measuring device only for media to which the process-wetted materials are
sufficiently resistant.

‣ Observe the limit values in the "Technical data".
Incorrect use
The manufacturer is not liable for damage caused by improper or non-intended use.
Avoid mechanical damage:‣ Do not touch or clean device surfaces with pointed or hard objects.
Clarification of borderline cases:‣ For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistancein verifying the corrosion resistance of fluid-wetted materials, but does not accept any
warranty or liability.

Residual risks
Due to heat transfer from the process as well as power loss in the electronics, thetemperature of the electronics housing and the assemblies it contains (e.g. display module,
main electronics module and I/O electronics module) may rise to 80 °C (176 °F). When in
operation, the sensor may reach a temperature close to the medium temperature.

8

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Basic safety instructions

Danger of burns from contact with surfaces!
‣ For elevated fluid temperatures, ensure protection against contact to prevent burns.

2.3

Workplace safety

When working on and with the device:‣ Wear the required personal protective equipment according to federal/national
regulations.
‣ Switch off the supply voltage before connecting the device.

2.4

Operational safety

Risk of injury!
‣ Operate the device only if it is in proper technical condition, free from errors and faults.
‣ The operator is responsible for the interference-free operation of the device.
Modifications to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers:‣ If modifications are nevertheless required, consult with the manufacturer.
RepairTo ensure continued operational safety and reliability:‣ Carry out repairs on the device only if they are expressly permitted.
‣ Observe federal/national regulations pertaining to the repair of an electrical device.
‣ Use only original spare parts and accessories from the manufacturer.
Hazardous area
To eliminate danger to persons or the installation when the device is used in thehazardous area (e.g. explosion protection, pressure vessel safety):‣ Check the nameplate to verify whether the ordered device can be put to its intended
use in the hazardous area.
‣ Observe the specifications in the separate supplementary documentation, which is an
integral part of this manual.

2.5

Product safety

This device is designed in accordance with good engineering practice to meet state-of-theart safety requirements, has been tested, and left the factory in a condition in which it issafe to operate.
It meets the general safety standards and legal requirements. It also complies with the EU
directives listed in the device-specific EU Declaration of Conformity. Endress+Hauser
confirms this by affixing the CE mark to the device.

2.6

IT security

We only provide a warranty if the device is installed and used as described in the
Operating Instructions. The device is equipped with security mechanisms to protect itagainst any inadvertent changes to the device settings. IT security measures in line with
operators' security standards and designed to provide additional protection for the device
and device data transfer must be implemented by the operators themselves.

Endress+Hauser

9Basic safety instructions

Micropilot FMR60B PROFIBUS PA

2.7

Device-specific IT security

The device offers specific functions to support protective measures by the operator. These
functions can be configured by the user and guarantee greater in-operation safety if usedcorrectly. An overview of the most important functions is provided in the following section:• Write protection via hardware write protection switch• Access code (applies for operation via display, Bluetooth or FieldCare, DeviceCare, PDM)

10Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Product description

3

Product description

3.1

Product design

1

2
4
3
5

A0046661

1
1
2
3
4
5

Endress+Hauser

Design of the Micropilot FMR60B
Electronics housing
Encapsulated antenna, PVDF, 40 mm (1.5 in)
Drip-off antenna 50 mm (2 in) - threaded or UNI flange process connection
Integrated antenna, PEEK, 20 mm (0.75 in)Integrated antenna, PEEK, 40 mm (1.5 in)

11Incoming acceptance and product identification

Micropilot FMR60B PROFIBUS PA

4

Incoming acceptance and productidentification

4.1

Incoming acceptance
DELIVERY NOTE

1=2

A0016870

Check the following during incoming acceptance:• Is the order code on the delivery note (1) identical to the order code on the productsticker (2)?
• Are the goods undamaged?
• Do the data on the nameplate correspond to the order specifications and the delivery
note?
• Is the documentation provided?
• If required (see nameplate): are the Safety Instructions (XA) provided?
If one of these conditions is not met, please contact the manufacturer's sales office.

4.2

Product identification

The following options are available for identification of the device:• Nameplate specifications• Extended order code with breakdown of the device features on the delivery note

‣ Device Viewer(www.endress.com/deviceviewer); manually enter the serial number
from the nameplate.
 All the information about the measuring device is displayed.

‣ Endress+Hauser Operations app; manually enter the serial number indicated on the
nameplate or scan the 2D matrix code on the nameplate.
 All the information about the measuring device is displayed.

4.2.1

Nameplate

The information that is required by law and is relevant to the device is shown on the
nameplate, e.g.:• Manufacturer identification• Order number, extended order code, serial number• Technical data, degree of protection• Firmware version, hardware version• Approval-related information, reference to Safety Instructions (XA)
• DataMatrix code (information about the device)

12

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

4.2.2

Incoming acceptance and product identification

Manufacturer address

Endress+Hauser SE+Co. KG
Hauptstraße 179689 Maulburg, Germany
Place of manufacture: See nameplate.

4.3

Storage and transport

4.3.1

Storage conditions• Use the original packaging• Store the device in clean and dry conditions and protect from damage caused by shocks
Storage temperature range
See Technical Information.

4.3.2

Transporting the product to the measuring point

LWARNING
Incorrect transport!
The housing or sensor can be damaged or pull off. Risk of injury!
‣ Transport the device to the measuring point in its original packaging or by the processconnection.
‣ Always secure lifting equipment (slings, eyes, etc.) at the process connection and neverlift the device by the electronic housing or sensor. Pay attention to the center of gravityof the device so that it does not tilt or slip unintentionally.

Endress+Hauser

13

MountingMicropilot FMR60B PROFIBUS PA

5

Mounting

5.1

General instructions

LWARNING
Loss of protection rating if the device is opened in a wet environment.
‣ Only open the device in a dry environment!
1.

Install the device or turn the housing so that the cable entries do not point upwards.
A0029263

2.

Always firmly tighten the housing cover and the cable entries.

3.

Counter-tighten the cable entries.

4.

A drip loop must be provided when laying the cables.

5.2

Mounting requirements

5.2.1

Internal vessel fittingsαA0031777

Avoid internal fittings (point level switches, temperature sensors, struts, vacuum rings,
heating coils, baffles etc.) inside the signal beam. Pay attention to the beam angle α.

14

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Mounting

5.2.2

Avoiding interference echoes

A0031813

Metal deflector plates, installed at an angle to scatter the radar signals, help prevent
interference echoes.

5.2.3

Vertical alignment of antenna axis

Align the antenna so that it is perpendicular to the product surface.
The maximum reach of the antenna can be reduced, or additional interference signals
can occur, if the antenna is not installed perpendicular to the product.

5.2.4

Radial alignment of the antenna

Based on the directional characteristic, radial alignment of the antenna is not necessary.

5.2.5

Optimization options

Mapping
Measurement can be optimized by electronically suppressing interference echoes.
See the Confirm distance parameter.

5.3

Mounting the device

5.3.1

Encapsulated antenna, PVDF 40 mm (1.5 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 50 mm• Maximum permissible torque: 30 Nm (22 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

Endress+Hauser

15

MountingMicropilot FMR60B PROFIBUS PA

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D
D

Hmax

50 to 80 mm (2 to 3.2 in)

600 mm (24 in)

80 to 100 mm (3.2 to 4 in)

1 000 mm (24 in)

100 to 150 mm (4 to 6 in)

1 250 mm (50 in)≥ 150 mm (6 in)

1 850 mm (74 in)øDIn the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

5.3.2

Drip-off antenna PTFE 50 mm (2 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 55 mm• Maximum permissible torque: 50 Nm (36 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D
D

Hmax

50 to 80 mm (2 to 3.2 in)

750 mm (30 in)

80 to 100 mm (3.2 to 4 in)

1 150 mm (46 in)

100 to 150 mm (4 to 6 in)

1 450 mm (58 in)≥150 mm (6 in)

2 200 mm (88 in)øDIn the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

16

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Mounting

5.3.3

Integrated antenna, PEEK 20 mm (0.75 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 36 mm• Maximum permissible torque: 50 Nm (36 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D

øD

D

Hmax

40 to 50 mm (1.6 to 2 in)

200 mm (8 in)

50 to 80 mm (2 to 3.2 in)

300 mm (12 in)

80 to 100 mm (3.2 to 4 in)

450 mm (18 in)

100 to 150 mm (4 to 6 in)

550 mm (22 in)≥ 150 mm (6 in)

850 mm (34 in)In the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

5.3.4

Integrated antenna, PEEK 40 mm (1.5 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 55 mm• Maximum permissible torque: 50 Nm (36 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D
ØD

Hmax

40 to 50 mm (1.6 to 2 in)

400 mm (16 in)

50 to 80 mm (2 to 3.2 in)

550 mm (22 in)

80 to 100 mm (3.2 to 4 in)

850 mm (34 in)øD

Endress+Hauser

17

MountingMicropilot FMR60B PROFIBUS PA

ØD

Hmax

100 to 150 mm (4 to 6 in)

1 050 mm (42 in)≥150 mm (6 in)

1 600 mm (64 in)In the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

5.3.5

Turning the housing

The housing can be rotated up to 380° by loosening the locking screw.
Your benefits• Easy installation due to optimum alignment of housing• Easily accessible device operation• Optimum readability of the local display (optional)
B

A

C

1

E

D

1

1
A0046660

A
B
C
D
E
1

Plastic single compartment housing (no locking screw)
Aluminum single compartment housing
Single compartment housing, 316L, hygiene (no locking screw)
Dual compartment housing
Dual compartment housing, L-form
Locking screw

NOTICEThe housing cannot be unscrewed fully.
‣ Loosen the external locking screw by a maximum of 1.5 turns. If the screw isunscrewed too much or completely (beyond the screw anchor point), small parts(counter disk) can become loose and fall out.
Tightenthe securing screw (hexagon socket 4 mm (0.16 in)) with maximum‣
3.5 Nm (2.58 lbf ft) ± 0.3 Nm (0.22 lbf ft).

5.3.6

Turning the display module

LWARNING
Supply voltage switched on!
Risk of electric shock and/or explosion!
‣ Switch off the supply voltage before opening the measuring device.

18

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Mounting

3.

1.
4 0.7 Nm

2.

4.

A0038224

1.

If fitted: release the screw of the cover lock for the electronics compartment coverusing the Allen key.

2.

Unscrew the electronics compartment cover from the transmitter housing and checkthe cover seal.

3.

Press the release mechanism and remove the display module.

4.

Turn the display module to the desired position: maximum 4 × 90 ° in each direction.
Fit the display module on the electronics compartment in the desired position until itclicks into place. Screw the electronics compartment cover back onto the transmitter
housing. If fitted: tighten the screw of the cover lock using the Allen key
0.7 Nm (0.52 lbf ft)±0.2 Nm (0.15 lbf ft).

5.3.7

Changing the installation position of the display module

The installation position of the display can be changed in the case of the dual
compartment housing, L-form.

A0048401

Endress+Hauser

19

MountingMicropilot FMR60B PROFIBUS PA

LWARNING
Supply voltage switched on!
Risk of electric shock and/or explosion!
‣ Switch off the supply voltage before opening the measuring device.
1→

2→

3→

4 mm

1.

2.
2.

2x

1.
0.7 Nm

A0046831

A0046832

A0046833‣ If fitted: release the screw of‣ Unscrew the display cover and‣ Press the release mechanism,

4→

5→

6→

the cover lock for the display
cover using the Allen key.

check the cover seal.

remove the display module.

4 mm
2x

1.
0.7 Nm2.
A0046834‣ Release the connection.

20

A0046923‣ If fitted: release the screw of

the cover lock for theconnection compartment coverusing the Allen key.

A0046924‣ Unscrew the connection

compartment cover, check thecover seal. Screw this cover
onto the electronics
compartment instead of thedisplay cover. If fitted: tighten
the screw of the cover lockusing the Allen key

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Mounting

7→

8

Di

sp

lay

2.

1.
A0048406‣ Plug in the connection for the
display module in theconnection compartment.
‣ Fit the display module in thedesired position until it clicks
into place.

5.3.8

A0046928‣ Screw the display cover firmly

back onto the housing. If fitted:
tighten the screw of the coverlock using the Allen key
0.7 Nm (0.52 lbf ft).

Closing the housing covers

NOTICE
Thread and housing damaged from dirt!
‣ Remove dirt (e.g. sand) on the cover and housing thread.
‣ If you encounter resistance when closing the cover, check the thread for dirt and cleanit.
Housing thread
The threads of the electronics and connection compartment can be coated with ananti-friction coating.
The following applies for all housing materials:
Do not lubricate the housing threads.

5.4

Post-mounting checkIs the device free from damage (visual inspection)?



Are the measuring point identification and labeling correct (visual inspection)?

Is the measuring device protected against precipitation and sunlight?



Are the securing screws and cover lock tightened securely?

 Does the measuring device comply with the measuring point specifications?
For example:•  Process temperature•  Process pressure•  Ambient temperature•  Measuring range

Endress+Hauser

21

Electrical connection

Micropilot FMR60B PROFIBUS PA

6

Electrical connection

6.1

Connecting requirements

6.1.1

Cover with securing screw

The cover is locked by a securing screw in devices for use in hazardous areas with certain
explosion protection.
NOTICE
If the securing screw is not positioned correctly, the cover cannot provide securesealing.
‣ Open the cover: slacken the screw of the cover lock with a maximum of 2 turns so thatthe screw does not fall out. Fit the cover and check the cover seal.
‣ Close the cover: screw the cover securely onto the housing, making sure that the
securing screw is positioned correctly. There should not be any gap between the coverand housing.

2x4
0.7 Nm

A0039520

2

Cover with securing screw

7x4
0.7 Nm

A0050983

3

6.1.2

Cover with securing screw; hygienic housing (only for dust explosion protection)

Potential equalization

The protective ground on the device must not be connected. If necessary, the potential
matching line can be connected to the outer ground terminal of the transmitter before thedevice is connected.

22

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Electrical connection

B

A

1

1

D

C

1

1

E

1
A0046583

A
B
C
D
E
1

Single compartment housing, plastic
Single compartment housing, aluminum
Single compartment housing, 316L hygiene (Ex device)
Dual compartment housing
Dual compartment housing, L-form
Ground terminal for connecting the potential matching line

LWARNING
Explosion Hazard!
‣ Please refer to the separate documentation on applications in hazardous areas for thesafety instructions.
For optimum electromagnetic compatibility:• Keep the potential matching line as short as possible• Observe a cross-section of at least 2.5 mm2 (14 AWG)

6.2

Connecting the device
C

B

A
1

1

D
1

E
1

1

A0046659

A
B
C
D
E
1

Plastic single compartment housing
Aluminum single compartment housing
Single compartment housing, 316L hygiene
Dual compartment housing
Dual compartment housing, L-form
Connection compartment coverDevices with a single compartment housing, 316L hygiene, and conduit gland must beconnected as end-of-line devices. Only one conduit entry must be used.
Housing thread
The threads of the electronics and connection compartment can be coated with ananti-friction coating.
The following applies for all housing materials:
Do not lubricate the housing threads.

6.2.1

Supply voltage

The supply voltage depends on the selected type of device approval

Endress+Hauser

Non-hazardous, Ex d, Ex e

9 to 32 VDC

Ex i

9 to 30 VDC

23

Electrical connection

Micropilot FMR60B PROFIBUS PA

Nominal current

14 mA

Failure current FDE (Fault
Disconnection Electronic)

0 mA• Use only suitable and certified Profibus PA components (e.g. DP/PA segment
coupler) for the power supply• FISCO/FNICO-compliant according to IEC 60079-27• The supply is not polarity-dependent
Device display and Bluetooth
The display background lighting and Bluetooth function (order option) is guaranteed
across the entire supply voltage range. The Bluetooth function can be restricted at highambient temperatures.

6.2.2

Cable specification

Rated cross-section• Supply voltage
0.5 to 2.5 mm2 (20 to 13 AWG)
• Protective earth or grounding of the cable shield> 1 mm2 (17 AWG)
• External ground terminal
0.5 to 4 mm2 (20 to 12 AWG)
Cable outer diameter
The cable outer diameter depends on the cable gland used
• Coupling, plastic:⌀5 to 10 mm (0.2 to 0.38 in)• Coupling, nickel-plated brass:⌀7 to 10.5 mm (0.28 to 0.41 in)• Coupling, stainless steel:⌀7 to 12 mm (0.28 to 0.47 in)
Use a twisted, shielded twin-core cable, preferably cable type A.
For further information on the cable specification:•
Operating Instructions BA00034S "PROFIBUS DP/PA: Guidelines for planningand commissioning"•
PROFIBUS Assembling Guideline 8.022•
IEC 61158-2 (MBP).

6.2.3

Overvoltage protection

The overvoltage protection can optionally be ordered as a "Mounted accessory" via the
product structure
Devices without optional overvoltage protection
The equipment fulfills the requirements of the product standard IEC / DIN EN 61326-1(Table 2 Industrial Environment).
Depending on the type of port (DC power supply, input/output port) different testing levels
according to IEC / DIN EN 61326-1 against transient overvoltages (Surge) are applied(IEC / DIN EN 61000-4-5 Surge):
Test level on DC power ports and input/output ports is 1 000 V line to earth

24

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Electrical connection

Devices with optional overvoltage protection• Spark-over voltage: min. 400 VDC• Tested according to IEC / DIN EN 60079-14 sub chapter 12.3 (IEC / DIN EN 60060-1
chapter 7)• Nominal discharge current: 10 kA
NOTICE
Device could be destroyed‣ Always ground device with integrated overvoltage protection.
Overvoltage category
Overvoltage category II

6.2.4

Wiring

LWARNING
Supply voltage might be connected!
Risk of electric shock and/or explosion!
‣ If the device is used in hazardous areas, make sure to comply with national standards
and the specifications in the Safety Instructions (XAs). The specified cable gland must
be used.
The
supply voltage must match the specifications on the nameplate.
‣
Switchoff the supply voltage before connecting the device.
‣Ifnecessary,
the potential matching line can be connected to the outer ground terminal‣of the device before the power supply lines are connected.
‣ FISCO/FNICO compliant according to IEC 60079-27.
‣ The supply does not depend on the polarity.
‣ The cables must be adequately insulated, with due consideration given to the supply
voltage and the overvoltage category.
‣ The connecting cables must offer adequate temperature stability, with due
consideration given to the ambient temperature.
‣ Only operate the measuring device with the covers closed.
Connect the device in the following order:

Endress+Hauser

1.

Release the cover lock (if provided).

2.

Unscrew the cover.

3.

Guide the cables into the cable glands or cable entries.

4.

Connect the cable.

5.

Tighten the cable glands or cable entries so that they are leak-tight. Counter-tighten
the housing entry.

6.

Screw the cover securely back onto the connection compartment.

7.

If provided: tighten the screw of the cover lock using the Allen key
0.7 Nm (0.52 lbf ft) ±0.2 Nm (0.15 lbf ft).

25

Electrical connection

Micropilot FMR60B PROFIBUS PA

6.2.5

Terminal assignment

Single compartment housing

3
2

-

1

+

A00425944123Connection terminals and ground terminal in the connection compartment
Positive terminal
Negative terminalInternal ground terminal

Dual compartment housing

- +

3

2 1
A0042803

5
1
2
3

26

Connection terminals and ground terminal in the connection compartment
Positive terminal
Negative terminalInternal ground terminal

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Electrical connection

Dual compartment housing, L-form

- +

3

2 1
A00458426123Connection terminals and ground terminal in the connection compartment
Positive terminal
Negative terminalInternal ground terminal

6.2.6

Cable entries
C

B

A
1

D

E

1

1

1

1

2

2

2

2

2
A0046584

A
B
C
D
E
1
2

Single compartment housing, plastic
Single compartment housing, aluminum
Single compartment housing, 316L hygiene
Dual compartment housing
Dual compartment housing, L-form
Cable entry
Dummy plug

The type of cable entry depends on the device version ordered.
Always route connecting cables downwards so that moisture cannot penetrate theconnection compartment.
If necessary, create a drip loop or use a weather protection cover.

6.2.7

Available connectors

In the case of devices with a plug, it is not necessary to open the housing forconnection purposes.
Use the enclosed seals to prevent the penetration of moisture into the device.

Endress+Hauser

27

Electrical connection

Micropilot FMR60B PROFIBUS PA

Devices with M12 plug

4

3

1

2
A0011175

7
1
2
3
4

View of the plug-in connection on the device
Signal +Not assigned
Signal –GroundVarious M12 sockets are available as accessories for devices with M12 plugs.

6.3

Ensuring the degree of protection

6.3.1

Cable entries• M20 coupling, plastic, IP66/68 NEMA TYPE 4X/6P• M20 coupling, nickel-plated brass, IP66/68 NEMA TYPE 4X/6P• M20 coupling, 316L, IP66/68 NEMA TYPE 4X/6P• M20 thread, IP66/68 NEMA Type 4X/6P• G1/2 thread, IP66/68 NEMA Type 4X/6P
If the G1/2 thread is selected, the device is delivered with an M20 thread as standard
and a G1/2 adapter is included with the delivery, along with the corresponding
documentation• NPT 1/2 thread, IP66/68 NEMA Type 4X/6P• Dummy plug transport protection: IP22, NEMA TYPE 2• M12 plug
• When housing is closed and connecting cable is plugged in: IP66/67, NEMA Type 4X• When housing is open or connecting cable is not plugged in: IP20, NEMA Type 1
NOTICEM12 plug: Loss of IP protection class due to incorrect installation!
‣ The degree of protection only applies if the connecting cable used is plugged in andscrewed tight.
The
degree of protection only applies if the connecting cable used is specified according‣
to IP67, NEMA Type 4X.
‣ The IP protection classes are only maintained if the dummy cap is used or the cable isconnected.

6.4

28

Post-connection check

Are the device or cables undamaged (visual inspection)?



Do the cables used comply with the requirements?

Do the mounted cables have strain relief?



Cable glands mounted, securely tightened and leak-tight?

Does the supply voltage correspond to the specifications on the nameplate?



Cover screwed down correctly?



Cover lock tightened correctly?

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Operation options

7

Operation options

7.1

Overview of operation options• Operation via operating keys and DIP switches on the electronic insert• Operation via optical operating keys on the device display (optional)• Operation via Bluetooth® wireless technology (with optional device display with
Bluetooth) with SmartBlue app or FieldXpert, DeviceCare• Operation via operating tool (Endress+Hauser FieldCare/DeviceCare, PDM etc.)

7.2

Operating keys and DIP switches on the PROFIBUS PA
electronic insert

off on1

off on2
Display

ON

01

1

CK S D A

2

3

4

5

6

7

08

8

4 3

A0050986

8
1
2
3
4

Operating keys and DIP switches on the PROFIBUS PA electronic insert
Operating key for reset password (for Bluetooth login and Maintenance user role)
DIP switch for address configuration
DIP switch with no function
DIP switch for locking and unlocking the device

The setting of the DIP switches on the electronic insert has priority over the settings
made via other operation methods (e.g. FieldCare/DeviceCare).

7.2.1

Hardware addressing
ON
98

DIP

1

2

3

4

5

6

7

8

2 + 8 = 10
A0053770

9

Example of hardware addressing; switch 8 is set to the "ON" position; switches 1 to 7 define the address.

1.

Set switch 8 to the "ON" position.

2.

Using switches 1 to 7, set the address as indicated in the table below.

The change of address takes effect after 10 seconds. The device is restarted.
Assignment of switch values

Endress+Hauser

Switch

1

2

3

4

5

6

7

Value in "ON" position

1

2

4

8

16

32

64

Value in "OFF" position

0

0

0

0

0

0

0

29

Operation options

Micropilot FMR60B PROFIBUS PA

7.2.2

Software addressing
ON
98

DIP

1

2

3

4

5

6

7

8
A0053771

 10

Example of software addressing; switch 8 is set to the "OFF" position; the address is defined in theoperating menu

1.

Set switch 8 to "OFF".

2.

The device restarts automatically and reports the PROFIBUS address stored in thedevice. The factory setting is PROFIBUS address 126 or the PROFIBUS address
ordered with order code "Marking", "Bus address" option.

3.

Configure the address via the operating menu: Application → Profibus→ Configuration → Device address

7.3

Structure and function of the operating menu

The differences between the structure of the operating menus of the local display and theEndress+Hauser FieldCare or DeviceCare operating tools can be summarized as follows:
The local display is suitable for configuring simple applications.
The operating tools (FieldCare, DeviceCare, SmartBlue, PDM etc.) can be used to configure
the parameters of wide-ranging applications.
Wizards help the user to commission the various applications. The user is guided throughthe individual configuration steps.

7.3.1

User roles and related access authorization

The two user roles Operator and Maintenance (as-delivered state) have different write
access to the parameters if a device-specific access code has been defined. This access code
protects the device configuration from unauthorized access.
If an incorrect access code is entered, the user obtains the access rights of the Operator
role.

7.4

Access to operating menu via local display

7.4.1

Device display (optional)

Possible to operate the optical operating keys through the cover. No need to open thedevice.
Functions:• Display measured values, also fault and notice messages• Background lighting, which switches from green to red in the event of an error• The device display can be removed for easier operation
The device display is also optionally available with Bluetooth® wireless technology.

30

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Operation options

ESC

-

+

E

1
A0039284 11Graphic display with optical operating keys (1)•  key• Navigate down in the picklist• Edit the numerical values or characters within a function•  key• Navigate up in the picklist• Edit the numerical values or characters within a function•  key• Change from main display to main menu
• Confirm entry• Jump to the next item
• Select a menu item and activate edit mode
• Unlock/lock the display operation• Press and hold the  key to display a short description of the selected parameter (ifavailable)•  key and  key (ESC function)• Exit edit mode for a parameter without saving the changed value• Menu at a selection level: pressing the keys simultaneously takes the user back up alevel in the menu
• Press and hold the keys simultaneously to return to the upper level

7.4.2

Operation via Bluetooth® wireless technology (optional)

Prerequisite• Device with device display including Bluetooth• Smartphone or tablet with Endress+Hauser SmartBlue app or PC with DeviceCare fromversion 1.07.05 or FieldXpert SMT70
The connection has a range of up to 25 m (82 ft). The range can vary depending on
environmental conditions such as attachments, walls or ceilings.
The operating keys on the display are locked as soon as the device is connected via
Bluetooth.
A flashing Bluetooth symbol indicates that a Bluetooth connection is available.
SmartBlue app
1.

Scan the QR code or enter "SmartBlue" in the search field of the App Store or GooglePlay.
A0039186

Endress+Hauser

31Operation options

Micropilot FMR60B PROFIBUS PA

2.

Start the SmartBlue app.

3.

Select device from livelist displayed.

4.

Login:
 Enter the user name: admin
Password: serial number of the device.

5.

Change the password after logging in for the first time!

Prerequisites
System requirements
The SmartBlue app is available to download for smartphones or tablets.
Please see the "App Store (Apple)" or "Google Play Store" for information regarding thecompatibility of the SmartBlue app with mobile terminals.
Initial password
The serial number of the device serves as the initial password when the connection isestablished for the first time.
Please note the following
If the Bluetooth display is removed from one device and installed in another device:• All the log-in data are only saved in the Bluetooth display and not in the device• The password changed by the user is also saved in the Bluetooth display

7.5

Access to the operating menu via the operating tool

Access via the operating tool is possible:• Via Profibus PA communication• Via Endress+Hauser Commubox FXA291
With the Commubox FXA291, a CDI connection can be established with the device
interface and a Windows PC/notebook with a USB port

7.6

DeviceCare

7.6.1

Function scope

Tool for connecting and configuring Endress+Hauser field devices.
The fastest way to configure Endress+Hauser field devices is with the dedicated"DeviceCare" tool. Together with the device type managers (DTMs), DeviceCare presents a
convenient, comprehensive solution.
For details, see Innovation Brochure IN01047S

7.7

FieldCare

7.7.1

Function range

FDT-based plant asset management tool from Endress+Hauser. FieldCare can configure all
smart field devices in a system and helps you manage them. By using the status
information, FieldCare is also a simple but effective way of checking their status andcondition.
Access is via:
• PROFIBUS PA protocol• CDI service interface

32

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Operation options

Typical functions:• Parameter configuration of transmitters• Loading and saving of device data (upload/download)• Documentation of the measuring point• Visualization of the measured value memory (line recorder) and event logbook
For additional information about FieldCare, see Operating Instructions BA00027S
and BA00059S

Endress+Hauser

33

System integration

Micropilot FMR60B PROFIBUS PA

8

System integration

According to EN 50170 Volume 2, IEC 61158-2 (MBP) type 1 PROFIBUS PA Profile
Version 3.02

8.1

Protocol-specific data

8.1.1

PROFIBUS PA

Manufacturer ID:
17 (0x11)
Ident number:
0x1568 or 0x9700
Profile version:
3.02
GSD file and version
Information and files available at:• www.endress.com
On the product page for the device: Documents/Software → Device drivers• www.profibus.com
Output values
Analog Input:• Level linearized• Distance• Volume• Terminal voltage• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index, optional (Guidance → Heartbeat Technology → Build-up detection→ Build-up index)• Foam index, optional (Diagnostics → Heartbeat Technology → Foam detection → Foam
index)
Digital Input:
Is only available if the "Heartbeat Verification + Monitoring" application package wasselected.
• 168 Build-up detected, optional (Guidance → Heartbeat Technology → Build-up
detection → 168 Build-up detected)• 952 Foam detected, optional (Guidance → Heartbeat Technology → Foam detection →952 Foam detected)
Input values
Analog Output:
Analog value from PLC to be indicated on the display

34

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

System integration

Supported functions• Identification & maintenance
Straightforward device identification on the part of the control system and nameplate• Automatic Ident Number adoption
GSD compatibility mode for generic profile 0x9700 "Transmitter with 1 Analog Input"• Physical layer diagnostics
Installation check of the PROFIBUS segment and device using terminal voltage andmessage monitoring• PROFIBUS upload/download
Reading and writing parameters is up to ten times faster with PROFIBUS upload/
download• Condensed status
Straightforward and self-explanatory diagnostic information by categorizing diagnostic
messages that occur

8.2

Device master file (GSD)

In order to integrate the field devices into the bus system, the PROFIBUS system needs a
description of the device parameters, such as output data, input data, data format, data
volume and supported transmission rate. These data are available in the general station
description (GSD) which is provided to the PROFIBUS Master when the communication
system is commissioned.
In addition, device bit maps, which appear as icons in the network structure, can also be
integrated.
With the Profile 3.0 device master file (GSD) it is possible to exchange field devices madeby different manufacturers without having to reconfigure.
Generally speaking, two different GSD versions are possible with Profile 3.0 and higher.
• Before configuring, the user must decide which GSD should be used to operate thesystem.
• The setting can be changed via a Class 2 master.

8.2.1

Manufacturer-specific GSD

This GSD guarantees the unrestricted functionality of the measuring device. Device-specificprocess parameters, functions and all device diagnostics are therefore available.
Manufacturer-specific GSD

ID numberFile name

PROFIBUS PA

0x1568EH3x1568.gsd

If the manufacturer-specific GSD should be used, this is specified in the Ident number
selector parameter by selecting the FMR6xB 0x1568 option.
Where to acquire the manufacturer-specific GSD:
www.endress.com → Download

8.2.2

Profile GSD

Differs in terms of the number of Analog Input blocks (AI) and the measured values. If asystem is configured with a Profile GSD, it is possible to exchange devices made bydifferent manufacturers. However, it is essential to ensure that the order of the cyclic
process values is correct.

Endress+Hauser

ID number

Supported blocks

0x9700

1 Analog Input

35

System integration

Micropilot FMR60B PROFIBUS PA

The Profile GSD that is to be used is specified in the Ident number selector parameter byselecting the 0x9700 (1AI) option or FMR6xB 0x1568 option.

8.2.3

Compatibility with other devices

This device ensures compatibility in cyclic data exchange with the automation system(Class 1 master) for the following devices:
Generic Transmitter 1 AI PROFIBUS PA (Profile version 3.02, ident number 0x9700)
It is possible to replace these devices without the need to reconfigure the PROFIBUS
network in the automation unit although the name and identification number of thedevices are different.
Once replaced, the device is either identified automatically (factory setting) or device
identification can be set manually.
Automatic identification (factory setting)
The device automatically recognizes the generic profile configured in the automation
system and makes the same input data and measured value status information available
for cyclic data exchange. Automatic identification is set in the Ident number selector
parameter using the Automatic mode option (factory setting).
Manual setting
The manual setting is made in the Ident number selector parameter via the FMR6xB
0x1568 option (manufacturer) or 0x9700 (1AI) option (generic).
Afterwards, the device makes the same input and output data and measured status
information available for cyclic data exchange.
• If the device is acyclically configured via an operating program (Class 2 master),access is directly via the block structure or the parameters of the device.
• If parameters were changed in the device to be replaced (parameter setting nolonger corresponds to the original factory setting), these parameters have to beadapted accordingly in the new device used via an operating program (Class 2
master) to guarantee identical behavior• The cyclic data exchange of the Micropilot FMR6xB is not downward compatible
with the Micropilot FMR5x or Micropilot FMR4x

8.3

Cyclic data transmission

Cyclic data transmission when using the device master file (GSD).

8.3.1

Block model

The block model shows which input and output data the measuring device makes available
for cyclic data exchange. The cyclic data exchange is performed with a PROFIBUS master(Class 1), e.g. a control system.
Transducer Block• Analog Input Block 1 to 6; AI output values →• Digital Input Block 1 to 2; DI output values →• Analog Output Block 1; AO input value ←Defined order of modules
The device works as a modular PROFIBUS slave. In contrast to a compact slave, a modular
slave has a variable design and consists of several individual modules. The device master

36

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

System integration

file (GSD) contains a description of the individual modules (input and output data) alongwith their individual properties.
The modules are permanently assigned to the slots, i.e. when configuring the modules, theorder and the arrangement of the modules must be respected.
Slot

ModuleFunction block

01 … 06

AI

Analog Input block 1 to 6

07 … 08

DI

Digital Input Block 1 to 2

09

DO

Digital Output Block 1

To optimize the data throughput rate of the PROFIBUS network, it is advisable to only
configure modules that are processed in the PROFIBUS master system. If this results ingaps between the configured modules, these gaps must be assigned to the
EMPTY_MODULE.

8.3.2

Description of the modules

The data structure is described from the perspective of the PROFIBUS master:• Input data: Are sent from the measuring device to the PROFIBUS master.
• Output data: Are sent to the measuring device by the PROFIBUS master.
AI module (Analog Input)
Transmit an input variable from the measuring device to the PROFIBUS master (Class 1).
The selected input variable, including the status, is cyclically transmitted to the PROFIBUS
master (Class 1) via the AI module. The input variable is depicted in the first four bytes in
the form of a floating point number as per the IEEE 754 standard. The fifth byte contains
standardized status information pertaining to the input variable.
Six Analog Input blocks are available (slot 1 to 6); Analog input 1 to 6 submenu
Input variable:
The input variable can be determined using the Channel parameter.
Application → Profibus → Analog input → Analog input 1 to 6 → Channel
Options:• Level linearized• Volume• Distance• Terminal voltage• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index option Visibility depends on order options or device settingsGuidance→ Heartbeat Technology → Build-up detection → Configuration → Build-up index• Foam index option Visibility depends on order options or device settingsGuidance→ Heartbeat Technology → Foam detection → Configuration → Foam index
Factory setting: Level linearized option
Data structure
Byte 1

Byte 2Byte 3

Byte 4

Measured value: Floating point number (IEEE 754)

Endress+Hauser

Byte 5
Status37

System integration

Micropilot FMR60B PROFIBUS PA

AO module (Analog Output)
Transmit display information from the PROFIBUS master (Class 1) to the measuring
device.
An analog output value, along with the status, can be transmitted cyclically from the
PROFIBUS master (Class 1) to the measuring device via the AO module and output on thelocal display. The value is displayed in the first four bytes in the form of a floating point
number as per the IEEE 754 standard. The fifth byte contains standardized status
information pertaining to the output value.
One Analog Output block is available (slot 9).
Application → Profibus → Analog output → Analog output 1 → Out valueOut value
parameter; Shows an analog output value (AO) that is output from the controller tothe device and can be shown on the local display. To show the AO on the local display,
it must be assigned to a display output parameter as a value. This assignment is madein the menu under "System-Display".
Data structure
Byte 1

Byte 2Byte 3

Byte 4

Measured value: Floating point number (IEEE 754)

Byte 5
StatusDI module (digital input)
Transmit discrete input values from the measuring device to the PROFIBUS master (Class
1). Discrete input values are used by the measuring device to transmit the state of device
functions to the PROFIBUS master (Class 1).
The DI module cyclically transmits the discrete input value, along with the status, to the
PROFIBUS master (Class 1). The discrete input value is depicted in the first byte. The
second byte contains standardized status information pertaining to the input value.
Two discrete input blocks are available (slot 7 to 8).
The digital input blocks are available only if the Heartbeat option is available and one
or both of the Heartbeat Monitoring functions Foam detected option and Build-up
detected option have been configured.
Device function:
The device function can be defined via the Channel parameter.
Application → Profibus → Digital input → Digital input 1 to 2 → Channel
Options:• Build-up index option Visibility depends on order options or device settingsGuidance→ Heartbeat Technology → Build-up detection → Configuration → Build-up index• Foam index option Visibility depends on order options or device settingsGuidance→ Heartbeat Technology → Foam detection → Configuration → Foam index
Factory setting: None option
Data structure
Byte 1

Byte 2

DigitalStatusEMPTY_MODULE moduleThis module is used to assign empty spaces arising from modules not being used in theslots.

38

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

System integration

The device works as a modular PROFIBUS slave. In contrast to a compact slave, a modular
PROFIBUS slave has a variable design and consists of several individual modules. The GSD
file contains a description of the individual modules along with their individual properties.
The modules are permanently assigned to the slots. When configuring the modules, it isabsolutely essential to observe the sequence/arrangement of the modules. Any gaps
between the configured modules must be filled with the EMPTY_MODULE.

Endress+Hauser

39Commissioning

Micropilot FMR60B PROFIBUS PA

9Commissioning
All configuration tools provide a commissioning assistant that supports the user whensetting the most important configuration parameters (Guidance menu
Commissioning wizard).

9.1

Preliminaries

The measuring range and the unit in which the measured value is transmitted correspondto the data on the nameplate.

9.2

Function check

Before commissioning the measuring point, check whether the post-installation and postconnection checks have been performed.
•→
•→

A
A

Post-installation check
Post-connection check

9.3

Connecting via FieldCare and DeviceCare

9.3.1

Via PROFIBUS PA protocol
2

3

PROFIBUS DP

1

PROFIBUS PA
T
5

4

4

4
A0050944

1
2
3
4
5

40

Segment coupler
Computer with PROFIusb and operating tool (e.g. DeviceCare/FieldCare )PLC (programmable logic controller)
Transmitter
Additional functions (valves etc.)

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Commissioning

9.3.2

Via service interface (CDI)1

2

3A00391481
2
3

Computer with FieldCare/DeviceCare operating toolCommubox FXA291
Service interface (CDI) of the measuring device (= Endress+Hauser Common Data Interface)

9.4

Setting the operating language

9.4.1

Local display

Setting the operating language
To set the operating language, the display first needs to be unlocked:
1.

Press the  key for at least 2 s.
 A dialog box appears.

2.

Unlock the display operation.

3.

Select the Language parameter in the main menu.

4.

Press the  key.

5.

Select the desired language with the  or  key.

6.

Press the  key.
Display operation locks automatically (except in the Safety mode wizard):• after 1 min on the main page if no key has been pressed• after 10 min within the operating menu if no key has been pressed

9.4.2

Operating toolSet display language
System → Display → Language
Selection in the Language parameter; Visibility depends on order options or device
settings

Endress+Hauser

41Commissioning

Micropilot FMR60B PROFIBUS PA

9.5

Configuring the device

9.5.1

Level measurement in liquids

A

R

C

L

F

E

D

100%

0%

A0016933

 12
R
A
C
D
L
E
F

Configuration parameters for level measurement in liquids

Reference point of measurement
Length of antenna + 10 mm (0.4 in)
50 to 80 mm (1.97 to 3.15 in); medium εr < 2
Distance
Level"Empty calibration" parameter (= 0 %)"Full calibration" parameter (= 100 %)In the case of media with a low dielectric constant, εr < 2 , the tank floor may be visible
through the medium at very low levels (lower than level C). Reduced accuracy must be
expected in this range. If this is not acceptable, the zero point should be positioned at a
distance C above the tank floor in these applications → Configuration parameters forlevel measurement in liquids.

9.5.2

Commissioning with the Commissioning wizard

In FieldCare, DeviceCare, SmartBlue and on the display, the Commissioning wizard isavailable to guide the user through the initial commissioning steps.
Complete this wizard to commission the device.
For each parameter, enter the appropriate value or select the appropriate option.
NOTE
If you exit the wizard before completing all required parameters, the changes you havemade will be saved. For this reason, the device may then be in an undefined state!
In this case, a reset to the default settings is recommended.

42

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

9.6

Commissioning

Recording the Echo curve

Record the current Echo curve as a reference echo curve for subsequent diagnostic
purposes
After the measurement has been configured, it is recommended to record the current Echo
curve as a reference echo curve.
The Save reference curve parameter in the Echo curve submenu is used to record the
Echo curve.
Diagnostics → Echo curve → Save reference curve‣ Under the Save reference curve parameter, activate the Customer reference curve
option9.7"Simulation" submenu

Simulation of a process variable, a pulse output or a diagnostic event.

Endress+Hauser

43

Operation

Micropilot FMR60B PROFIBUS PA

10Operation

10.1

Reading off the device locking status

Displaying active write protection in the Locking status parameter:• Local display
The symbol appears on the main page
• Operating tool (FieldCare/DeviceCare) :Navigation: System → Device management → Locking status

10.2

Reading off measured values

All the measured values can be read off using the Measured values submenu.
Navigation: Application menu → Measured values submenu

10.3

Adapting the device to process conditions

The following menus are available for this purpose:• Basic settings in the Guidance menu
• Advanced settings in:• Diagnostics menu
• Application menu
• System menu

10.4

Heartbeat Technology (optional)

10.4.1

Heartbeat Verification"Heartbeat Verification" wizard
This wizard is used to start an automatic verification of the device functionality. The
results can be documented as a verification report.
• The wizard can be used via the operating tools and the display
The wizard can be started on the display but only shows the Passed option or Failed
option result.
• The wizard guides the user through the entire process for creating the verification report

10.4.2

Heartbeat Verification/Monitoring

The Heartbeat submenu is only available if operating via FieldCare, DeviceCare or the
SmartBlue app. It contains the wizards that are available with the application
packages Heartbeat Verification and Heartbeat Monitoring.
Documentation on the Heartbeat Technology: Endress+Hauser website:www.endress.com → Downloads.

44

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Diagnostics and troubleshooting

11

Diagnostics and troubleshooting

11.1

General troubleshooting

11.1.1

General errors

Device is not responding• Possible cause: Supply voltage does not match the specification on the nameplate
Remedial action: Apply the correct voltage• Possible cause: The connecting cables are not in contact with the terminals
Remedial action: Check the electrical contact between cables and correct if necessary
Values not visible on the display• Possible cause: Display setting is too bright or too dark
Remedy:Use the Contrast display parameter to increase or reduce contrast
Navigation path: System → Display → Contrast display• Possible cause: The plug of the display cable is not connected correctly
Remedial action: Connect the plug correctly• Possible cause: Display is defective
Remedial action: Replace the display"Communication error" is indicated on the display when the device is started or the
display is connected• Possible cause: Electromagnetic interference influence
Remedial action: Check grounding of the device• Possible cause: Defective cable connection or display plug
Remedial action: Replace the display
Communication via CDI interface not working
Possible cause: Wrong setting of the COM port on the computer
Remedial action: Check the setting of the COM port on the computer and correct it ifnecessary
Device measuring incorrectly
Possible cause: Parameter configuration error
Remedial action: Check and correct the parameter configuration

11.2

Error - SmartBlue operation

Operation via SmartBlue is only possible on devices that have a display with Bluetooth(optionally available).
Device is not visible in the live list
• Possible cause: No Bluetooth connection available
Remedial action: Enable Bluetooth in the field device via display or software tool and/orin the smartphone/tablet• Possible cause: Bluetooth signal outside range
Remedial action: Reduce distance between field device and smartphone/tablet
The connection has a range of up to 25 m (82 ft).
Operating radius with intervisibility 10 m (33 ft)• Possible cause: Geopositioning is not enabled on Android devices or is not permitted for
the SmartBlue app
Remedial action: Enable/disable geolocation service on Android device for the SmartBlue
app• Display does not have Bluetooth• Historom is not connected

Endress+Hauser

45

Diagnostics and troubleshooting

Micropilot FMR60B PROFIBUS PA

Device appears in the live list but a connection cannot be established• Possible cause: The device is already connected with another smartphone/tablet via
Bluetooth
Only one point-to-point connection is permitted
Remedial action: Disconnect the smartphone/tablet from the device• Possible cause: Incorrect user name and password
Remedial action: The standard user name is "admin" and the password is the device serial
number indicated on the device nameplate (only if the password was not changed by theuser beforehand)
If the password has been forgotten, contact Endress+Hauser Service(www.addresses.endress.com)
Connection via SmartBlue not possible• Possible cause: Incorrect password entered
Remedial action: Enter the correct password, paying attention to lower/upper case
• Possible cause: Forgotten password
Remedial action: Contact Endress+Hauser Service (www.addresses.endress.com)
Login via SmartBlue not possible• Possible cause: Device is being put into operation for the first time
Remedial action: Enter the user name "admin" and the password (device serial number),paying attention to lower/upper case
• Possible cause: The electric current and voltage are not correct.
Remedial action: Increase the supply voltage.
Device cannot be operated via SmartBlue• Possible cause: Incorrect password entered
Remedial action: Enter the correct password, paying attention to lower/upper case
• Possible cause: Forgotten password
Remedial action: Contact Endress+Hauser Service (www.addresses.endress.com)• Possible cause: Operator option has no authorization
Remedial action: System → User management → User role Change to Maintenanceoption11.3

Diagnostic message

Faults detected by the self-monitoring system of the device are displayed as a diagnostic
message in alternation with the measured value.
XXXXXXXXX

S

20.50
X

XX

S

XXXXXXXXX

S801
Supply voltageiMenu
A0051136

 13

Diagnostic message displayed in alternating sequence with the measured value

If two or more diagnostic events occur simultaneously, only the diagnostic message withthe highest priority is shown.

46

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Diagnostics and troubleshooting

11.3.1

Diagnostic information on local display

1

X

2
A0043759 14
1
2

Segment display without keys

Status symbol for event level
Status signal with diagnostic event

XXXXXXXXXSS801
XXXXXXXXX

1
2
3i
A0043103

 15
1
2
3

Graphic display with keys

Status signal
Status symbol with diagnostic event and preceding symbol for event level
Event text

11.3.2

Diagnostic information in the operating tool

If a diagnostic event has occurred in the device, the status signal appears in the top left
status area of the operating tool together with the corresponding symbol for the event
level according to NAMUR NE 107.
Click the status signal to see the detailed status signal.
Pending diagnostic messages can also be displayed in the Active diagnostics parameter.
The diagnostic events and remedial measures can be printed out in the Diagnostic list
submenu.

11.3.3

Status signal

F
Failure (F)A device error has occurred. The measured value is no longer valid.
C
Function check (C)The device is in the service mode (e.g. during a simulation).
S
Out of specification (S)Device operation:• Outside of the technical specifications (e.g. during startup or a cleaning)• Outside of the configuration performed by the user (e.g. sensor frequency outside theconfigured span)
M
Maintenance required (M)Maintenance required. The measured value is still valid.
Endress+Hauser

47

Diagnostics and troubleshooting

Micropilot FMR60B PROFIBUS PA

11.3.4

Diagnostic event and event text

The fault can be identified by means of the diagnostic event. The event text helps you byproviding information about the fault. In addition, the associated status symbol isdisplayed in front of the diagnostic event.
2
1

Diagnostic list
Diagnostics 1

S

3

4
5
6

S801 Supply voltage

Diagnostics 2
Diagnostics 3

7
Supply voltage(ID:203)
S801 0d00h02m25s
Increase supply voltage

8A0051137

1
2
3
4
5
6
7
8

Diagnostic message
Symbol for event level
Status signal
Short text
Symbol for event level, status signal, diagnostic number
Remedial measureService ID
Operating time of occurrence

Symbol for event level
 "Alarm" status
Measurement is interrupted. The signal outputs adopt the defined alarm state. A
diagnostic message is generated.
 "Warning" status
The device continues to measure. A diagnostic message is generated.

11.4

Remedy informationCalling up

11.4.1Graphic display with keys

1→S

XXXXXXXXX

S801
Supply voltageiMenu

A0051131‣ Open the Diagnostic list submenu

48

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Diagnostics and troubleshooting

2→

Diagnostic list
Diagnostics 1

S

S801 Supply voltage

Diagnostics 2
Diagnostics 3

A0051132‣ Select the diagnostic event and confirm
3

Supply voltage(ID:203)
S801 0d00h02m25s
Increase supply voltage

A0051133‣ Remedy informationClosing

11.4.2

Operating menu

In the Diagnostic list submenu, up to 5 currently pending diagnostic messages can bedisplayed. If more than 5 messages are pending, the messages with the highest priority are
shown on the display.
Navigation path
Diagnostics → Diagnostic list
Queued diagnostic messages are also displayed in the Active diagnostics parameter.
Navigation: Diagnostics → Active diagnostics

11.5

Adapting the diagnostic information

The event level can be configured:
Navigation: Diagnostics → Diagnostic settings → Configuration

11.6

List of diagnostic events

If Remedy information Contact the Service Department is displayed(www.addresses.endress.com), be ready with the service ID shown.
Diagnostic
numberShort text

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

Diagnostic of sensor

Endress+Hauser

062

Sensor connection
faultyCheck sensor connection

151

Sensor electronic failure 1. Restart device
2. Contact service

F

Alarm

F

Alarm

49

Diagnostics and troubleshooting

Micropilot FMR60B PROFIBUS PA

Diagnostic
number

168

Short text

Build-up detected

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

Check process conditions

M

Warning 1)

Diagnostic of electronic242

Firmware incompatible

1. Check software
2. Flash or change main electronic
module

F

Alarm

252

Module incompatible

1. Check if correct electronic
module is plugged
2. Replace electronic module

F

Alarm

270

Main electronics
defective

Replace main electronics

F

Alarm

272

Main electronics faulty

1. Restart device
2. Contact service

F

Alarm

273

Main electronics
defective

Replace main electronics

F

Alarm

282

Data storage
inconsistent

Restart device

F

Alarm

283

Memory content
inconsistent

1. Restart device
2. Contact service

F

Alarm

287

Memory content
inconsistent

1. Restart device
2. Contact service

M

Warning388

Electronics andHistoROM defective

1. Restart device
2. Replace electronics andHistoROM
3. Contact service

F

Alarm

Diagnostic of configuration

50

410

Data transfer failed

1. Retry data transfer
2. Check connection

F

Alarm

412

Processing download

Download active, please waitCWarning435

Linearization faulty

Check linearization table

F

Alarm

437

Configuration
incompatible

1. Update firmware
2. Execute factory reset

F

Alarm

438

Dataset different

1. Check dataset file
2. Check device parameterization
3. Download new device
parameterization

M

Warning

482

Block in OOS

Set Block in AUTO mode

F

Alarm

484

Failure mode simulation Deactivate simulation
active

C

Alarm

485

Process variable
simulation active

Deactivate simulationC

Warning495

Diagnostic event
simulation active

Deactivate simulation

S

Warning

497

Block output simulation
activeDeactivate simulationC

Warning538

Configuration Sensor
Unit invalid

1. Check sensor configuration
2. Check device configuration

F

Alarm

585

Simulation distance

Deactivate simulationC

Warning

586

Record map

Recording of mapping
please waitCWarningEndress+Hauser

Micropilot FMR60B PROFIBUS PA

Diagnostics and troubleshooting

Diagnostic
numberShort text

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

Diagnostic of process

1)

801

Supply voltage too low

Increase supply voltage

F

Alarm

802

Supply voltage too highDecrease supply voltage

S

Warning825Electronics temperature 1. Check ambient temperature
2. Check process temperatureS

Warning826

Sensor temperature outof range

1. Check ambient temperature
2. Check process temperatureS

Warning

941

Echo lost

Check parameter "DC value"S

Warning 1)

942

In safety distance

1. Check level
2. Check safety distance
3. Reset self holding

S

Warning 1)

952

Foam detected

Check process conditions

S

Warning 1)

968

Level limited

1. Check level
2. Check limit parameters

S

WarningDiagnostic behavior can be changed.

11.7

Event logbook

11.7.1

Event history

A chronological overview of the event messages that have occurred is provided in the
Event logbook submenu. This submenu only exists if operating via the local display withkeys. In the case of operation via FieldCare, the event list can be displayed with the "Event
List / HistoROM" functionality of FieldCare.
Navigation:
Diagnostics → Event logbook → Event list
A maximum of 100 event messages can be displayed in chronological order.
The event history includes entries for:• Diagnostic events• Information events
In addition to the operating time when the event occurred, each event is also assigned asymbol that indicates whether the event has occurred or is finished:• Diagnostic event• : Occurrence of the event• : End of the event• Information event
: Occurrence of the event

Calling up and closing the remedial measures

Endress+Hauser

1.

Press .
 The message for the remedial measures for the selected diagnostic event opens.

2.

Press  +  simultaneously.
 The message about the remedial measures closes.

51

Diagnostics and troubleshooting

Micropilot FMR60B PROFIBUS PA

11.7.2Filtering the event logbook

Filters can be used to determine which category of event messages is displayed in the
Event list submenu.
Navigation: Diagnostics → Event logbook
Filter categories• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information

11.7.3

Overview of information events

Info number

52

Info name

I1000--------(Device ok)

I1079

Sensor changed

I1089

Power on

I1090

Configuration reset

I1091

Configuration changed

I11074

Device verification active

I1110

Write protection switch changed

I1151

History reset

I1154

Reset terminal voltage min/max

I1155

Reset electronics temperature

I1157

Memory error event list

I1256

Display: access status changed

I1335

Firmware changed

I1397

Fieldbus: access status changed

I1398

CDI: access status changed

I1440

Main electronic module changed

I1444

Device verification passed

I1445

Device verification failed

I1461

Sensor verification failed

I1512

Download started

I1513

Download finished

I1514

Upload started

I1515

Upload finished

I1551

Assignment error fixed

I1552

Failed: Main electronic verification

I1556

Safety mode off

I1956

Reset

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Diagnostics and troubleshooting

11.8

Resetting the device

11.8.1

Reset password via operating toolEnter a code to reset the current "Maintenance" password.
The code is delivered by your local support.
Navigation: System → User management → Reset password → Reset password
For detailed information on the Reset password parameter: Description of device
parameters.

11.8.2

Reset device via operating tool

Reset the device configuration - either entirely or in part - to a defined state
Navigation: System → Device management → Reset device
For detailed information on the Reset device parameter: Description of device
parameters.

11.8.3

Resetting the device via keys on the electronic insert

Resetting the password
15 s
I

I

I

I

Display

off onI

2×
A0050210

 16

Sequence for resetting the password

Delete/reset the password
1.

Press operating key I three times.
 The reset password function is started, the LED flashes.

2.

Press operating key I once within 15 s.
 The password is reset, the LED flashes briefly.

If operating key I is not pressed within 15 s, the action is canceled and the LED is no longer
lit.

Endress+Hauser

53

Diagnostics and troubleshooting

Micropilot FMR60B PROFIBUS PA

Resetting the device to the factory setting
12 s

off onI
DisplayI

50 ms
A0050015

 17

Sequence - Factory setting

Reset the device to the factory setting‣ Press operating key I for at least 12 s.
 Device data are reset to the factory setting, the LED flashes briefly.

11.9

Device information

All device information is contained in the Information submenu.
Navigation: System → Information
For detailed information on the Information submenu: Description of device
parameters.

11.10 Firmware history
The firmware version can explicitly be ordered via the product structure. This makes it
possible to ensure the compatibility of the firmware version with an existing orplanned system integration.
Version01.00.00• Initial software• Valid from: 11.01.2023

54

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Maintenance

12

MaintenanceNo special maintenance work is required.

12.1

Exterior cleaning

Notes on cleaning• The cleaning agents used should not corrode the surfaces and the seals• Observe the degree of protection of the device

12.2

Seals

The process seals, located on the process connection of the device, should be replaced
periodically. The interval between changes depends on the frequency of the cleaning
cycles, the cleaning temperature and the medium temperature.

Endress+Hauser

55

Repair

Micropilot FMR60B PROFIBUS PA

13

Repair

13.1

General information

13.1.1

Repair concept

Endress+Hauser repair concept• The devices have a modular design• Repairs can be carried out by Endress+Hauser Service or by appropriately trained
customers• Spare parts are grouped into logical kits with the associated replacement instructions
For more information on service and spare parts, please contact your Endress+Hauser
sales representative.

13.1.2

Repairs to Ex-approved devices

LWARNING
Incorrect repair can compromise electrical safety!
Explosion hazard!
‣ Only specialist personnel or the manufacturer's service team may carry out repairs on
Ex-certified devices in accordance with national regulations.
‣ Relevant standards and national regulations on hazardous areas, safety instructions
and certificates must be observed.
‣ Only use original spare parts from the manufacturer.
‣ Please note the device designation on the nameplate. Only identical parts may be used
as replacements.
‣ Carry out repairs according to the instructions.
‣ Only the manufacturer's service team is permitted to modify a certified device andconvert it to another certified version.

13.2

Spare parts• Some replaceable device components are identified by a spare part nameplate. This
contains information about the spare part.
• All the spare parts for the measuring device, along with the order code, are listed in the
Device Viewer (www.endress.com/deviceviewer) and can be ordered. If available, users
can also download the associated Installation Instructions.
Device serial number or QR code:
Located on the device and spare part nameplate.

13.3

Replacement

After an entire device or an electronics module has been replaced, the parameters can be
downloaded to the device again via the communication interface. For this, the data musthave been uploaded to the PC beforehand using the FieldCare/DeviceCare software.

13.3.1

HistoROM

It is not necessary to perform a new device calibration after replacing the display ortransmitter electronics.
The spare part is supplied without HistoROM.
After removing the transmitter electronics, remove HistoRom and insert it into the
new spare part.
56

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Repair

13.4

ReturnThe requirements for safe device return can vary depending on the device type andnational legislation.
1.

Refer to the web page for information:
http://www.endress.com/support/return-material
 Select the region.

2.

Return the device if repairs or a factory calibration are required, or if the wrong
device was ordered or delivered.

13.5

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize thedisposal of WEEE as unsorted municipal waste. Do not dispose of products bearing thismarking as unsorted municipal waste. Instead, return them to the manufacturer for
disposal under the applicable conditions.

Endress+Hauser

57

Accessories

Micropilot FMR60B PROFIBUS PA

14

Accessories

14.1

Weather protection cover 316L

The weather protection cover can be ordered together with the device via the "Accessory
enclosed" product structure.
It is used to protect against direct sunlight, precipitation and ice.
Weather protection cover 316L is suitable for the dual compartment housing made ofaluminum or 316L. The delivery includes the holder for direct mounting on the housing.
228.9 (9.01)
136.4 (5.37)
92.5 (3.64)

135 (5.31)

102.5 (4.04)

73.5 (2.89)

170.5 (6.71)!65 (2.5
6)
81 (3.19)
103 (4.06)
A0039231

 18

Dimensions. Unit of measurement mm (in)Material• Weather protection cover: 316L
• Clamping screw: A4• Holder: 316L
Order number for accessories:
71438303

14.2

Plastic weather protection cover

The weather protection cover can be ordered together with the device via the "Accessory
enclosed" product structure.
It is used to protect against direct sunlight, precipitation and ice.
The plastic weather protection cover is suitable for the single compartment housing madeof aluminum. The delivery includes the holder for direct mounting on the housing.

122 (4.8)!115 (4.53)

140 (5.51)
165 (6.5)

32 (1.26)
140 (5.51)
A0038280

 19

58

Dimensions. Unit of measurement mm (in)Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Accessories

Material
Plastic
Order number for accessories:
71438291

14.3

Mounting bracket, adjustable

The device can be mounted on a wall or ceiling with the mounting bracket.
The device can be aligned with the product surface with the swivel function.
The mounting bracket can be ordered together with the device via the product structure"Accessory enclosed".
Suitable for device with single compartment housing or dual compartment aluminum
housing, L-form, in combination with encapsulated antenna, PVDF, 40 mm (1.5 in) or
drip-off antenna 50 mm (2 in) with threaded process connection.

A0048745

 20

Ceiling or wall mounting

There is no conductive connection between the mounting bracket and the transmitter
housing. The bracket should be included in local potential equalization to prevent any
electrostatic charge.
Fasten only to stable materials (e.g. metal, brick, concrete) with suitable fastening
fixtures (provided by the customer).
Order number for accessories:
71597288

14.3.1

Dimensions

3 (0.12)9 (0.35)!9 (0.35)

35 (1.38)

6 (0.24)156 (6.14)

65 (2.56)
32.5 (1.28)

282 (11.1)
230 (9.06)

A0048769

 21

Endress+Hauser

Dimensions of mounting bracket. Unit of measurement mm (in)59Accessories

Micropilot FMR60B PROFIBUS PA

14.3.2

Scope of delivery

5
A0049050

 22

Scope of delivery of mounting bracket, adjustable

1 × mounting bracket, 316L (1.4404)
2 × holder, 316L (1.4404)
6 × screws, A4
4 × lock washer, A4

14.4

Remote display FHX50B

The remote display is ordered via the Product Configurator.
If the remote display is to be used, the device version Prepared for display FHX50B must
be ordered.
A

D

E
B

F
C

G

A0046692

A
B
C
D
E
F
G

60

Plastic single compartment housing, remote display
Aluminum single compartment housing, remote display
Single compartment housing, 316L hygiene, remote display
Device side, plastic single compartment housing prepared for display FHX50B
Device side, aluminum single compartment housing prepared for display FHX50B
Device side, dual compartment housing, L-form, prepared for display FHX50B
Device side, single compartment housing, 316L hygiene, prepared for display FHX50B

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Accessories

Material of single compartment housing, remote display• Aluminum• Plastic
Degree of protection:• IP68 / NEMA 6P• IP66 / NEMA 4x
Connecting cable:• Connecting cable (option) up to 30 m (98 ft)• Customer-supplied standard cable up to 60 m (197 ft)
Recommendation: EtherLine®-P CAT.5e from LAPP.
Specification of customer-supplied connecting cablePush-in CAGE CLAMP®, connection technology, push actuation• Conductor cross-section:• Solid conductor 0.2 to 0.75 mm2 (24 to 18 AWG)
• Fine-stranded conductor 0.2 to 0.75 mm2 (24 to 18 AWG)
• Fine-stranded conductor; with insulated ferrule 0.25 to 0.34 mm2• Fine-stranded conductor; without insulated ferrule 0.25 to 0.34 mm2• Stripping length 7 to 9 mm (0.28 to 0.35 in)• Outer diameter: 6 to 10 mm (0.24 to 0.4 in)• Maximum cable length: 60 m (197 ft)
Ambient temperature:• –40 to +80 °C (–40 to +176 °F)• Option: –50 to +80 °C (–58 to +176 °F)

14.5

Gas-tight feedthroughChemically inert glass feedthrough, which prevents gases from entering the electronics
housing.
Can optionally be ordered as "Accessory mounted" via the product structure.

14.6

Field Xpert SMT70

Universal, high-performance tablet PC for device configuration in Ex Zone 2 and non-Ex
areas
For details, see "Technical Information" TI01342S

14.7

DeviceCare SFE100

Configuration tool for HART, PROFIBUS and FOUNDATION Fieldbus field devices
Technical Information TI01134S

14.8

FieldCare SFE500

FDT-based plant asset management tool
It can configure all smart field units in your system and helps you manage them. By using
the status information, it is also a simple but effective way of checking their status andcondition.
Technical Information TI00028S

Endress+Hauser

61

Accessories

Micropilot FMR60B PROFIBUS PA

14.9

RID14

8-channel field indicator for fieldbus systems

40

50

30
20
10

0

A0011631

Displays 8 process- or calculated values for FOUNDATION Fieldbus™ or PROFIBUS® PA
protocolTechnical Information TI00145R and Operating Instructions BA01267K

14.10 RID16
8-channel field indicator for fieldbus systems

A0011634

Displays 8 process- or calculated values for FOUNDATION Fieldbus™ or PROFIBUS® PA
protocolTechnical Information TI00146R and Operating Instructions BA00284R

62

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Accessories

14.11 Fieldgate SFG500
Smart Ethernet/PROFIBUS gateway

Power

PB DP

RS485

Run

PB ErrLAN 1

Failure

LAN 2
Reset

LAN 2

LAN 1

SFG500A0028262

Parallel access to PROFIBUS networks, PROFIBUS and HART device status monitoring
Ethernet gateway basic mode with integrated web server and adaptive PROFIBUS master,Class 2 for communication with PROFIBUS devices.
Order number for accessories:
71116672
Operating Instructions BA01579S

Endress+Hauser

63Technical data

Micropilot FMR60B PROFIBUS PA

15

Technical data

15.1

Input

Measured variable

The measured variable is the distance between the reference point and the productsurface. The level is calculated based on "E", the empty distance entered.

Measuring range

The measuring range starts at the point where the beam hits the tank floor. Levels below
this point cannot be measured, particularly in the case of spherical bases or conical outlets.
Maximum measuring range
The maximum measuring range depends on the antenna size and design.
Antenna

Maximum measuring range

Encapsulated, PVDF, 40 mm (1.5 in)

40 m (131 ft)

Drip-off, PTFE, 50 mm (2 in)

50 m (164 ft)

Integrated, PEEK, 20 mm (0.75 in)

10 m (32.8 ft)

Integrated, PEEK, 40 mm (1.5 in)

22 m (72 ft)

Usable measuring range
The usable measuring range depends on the antenna size, the medium's reflective
properties, the installation position and any possible interference reflections.
In principle, measurement is possible up to the tip of the antenna.
To avoid any material damage from corrosive or aggressive media or deposit buildup on
the antenna, the end of the measuring range should be selected 10 mm (0.4 in) before thetip of the antenna.

64

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

A

R

0%

C

B

H

100%

A0051658

 23
A
B
C
H
R

Usable measuring range

Length of antenna + 10 mm (0.4 in)
Usable measuring range
50 to 80 mm (1.97 to 3.15 in); medium εr < 2
Vessel height
Reference point of measurement, varies depending on the antenna system

For further information on the reference point, see →A

Mechanical construction.

In the case of media with a low dielectric constant, εr < 2 , the tank floor may be visible
through the medium at very low levels (lower than level C). Reduced accuracy must be
expected in this range. If this is not acceptable, the zero point should be positioned at a
distance C above the tank floor in these applications → Usable measuring range.
The media groups and the possible measuring range are described as a function of theapplication and media group in the following section. If the dielectric constant of themedium is not known, to ensure a reliable measurement assume the medium belongs togroup B.

Endress+Hauser

65Technical data

Micropilot FMR60B PROFIBUS PA

Media groups• A0 (εr 1.2 to 1.4)
e.g. n-butane, liquid nitrogen, liquid hydrogen• A (εr 1.4 to 1.9)
Non-conductive liquids, e.g. liquefied gas• B (εr 1.9 to 4)
Non-conductive liquids, e.g. gasoline, oil, toluene, etc.
• C (εr 4 to 10)
e.g. concentrated acid, organic solvents, ester, aniline, etc.
• D (εr >10)
Conductive liquids, aqueous solutions, diluted acids, bases and alcohol
Measurement of the following media with absorbing gas phase
For example:• Ammonia• Acetone• Methylene chloride• Methyl ethyl ketone• Propylene oxide• VCM (vinyl chloride monomer)
To measure absorbing gases, either use a guided radar, measuring devices withanother measuring frequency or another measuring principle.
If measurements must be performed in one of these media, please contact Endress+Hauser.
For the dielectric constants (DC values) of many media commonly used in industry,please refer to:• Dielectric constant (DC value) Compendium CP01076F• The Endress+Hauser "DC Values app" (available for Android and iOS)
Measurement in storage vessel
Storage vessel - measuring conditions
Calm medium surface (e.g. bottom filling, filling via immersion tube or rare filling fromabove)
Integrated antenna, PEEK, 20 mm (0.75 in) in storage vessel

66

Media group

Measuring range

A0 (εr 1.2 to 1.4)

1.5 m (5 ft)

A (εr 1.4 to 1.9)

2.5 m (8 ft)

B (εr 1.9 to 4)

5 m (16 ft)

C (εr 4 to 10)

8 m (26 ft)

D (εr >10)

10 m (33 ft)

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Integrated antenna, PEEK, 40 mm (1.5 in) in storage vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

3 m (10 ft)

A (εr 1.4 to 1.9)

6 m (20 ft)

B (εr 1.9 to 4)

11 m (36 ft)

C (εr 4 to 10)

15 m (49 ft)

D (εr >10)

22 m (72 ft)

Encapsulated antenna, PVDF, 40 mm (1.5 in) in storage vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

7 m (23 ft)

A (εr 1.4 to 1.9)

12 m (39.4 ft)

B (εr 1.9 to 4)

23 m (75.5 ft)

C (εr 4 to 10)

40 m (131 ft)

D (εr >10)40 m (131 ft)

PTFE Drip-off antenna, 50 mm (2 in) in storage vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

7 m (23 ft)

A (εr 1.4 to 1.9)

12 m (39 ft)

B (εr 1.9 to 4)

23 m (75 ft)

C (εr 4 to 10)

40 m (131 ft)

D (εr >10)50 m (164 ft)

Measurement in buffer vessel
Buffer vessel - measuring conditions
Moving medium surface (e.g. permanent free filling from above, mixing jets)

Endress+Hauser

67Technical data

Micropilot FMR60B PROFIBUS PA

Integrated antenna, PEEK, 40 mm (1.5 in) in buffer vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

1.5 m (5 ft)

A (εr 1.4 to 1.9)

3 m (10 ft)

B (εr 1.9 to 4)

6 m (20 ft)

C (εr 4 to 10)

13 m (43 ft)

D (εr >10)20 m (66 ft)

Encapsulated antenna, PVDF, 40 mm (1.5 in) in buffer vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)4 m (13 ft)

A (εr 1.4 to 1.9)

7.5 m (24.6 ft)

B (εr 1.9 to 4)

15 m (49.2 ft)

C (εr 4 to 10)

25 m (82 ft)

D (εr >10)35 m (114.8 ft)

PTFE Drip-off antenna, 50 mm (2 in) in buffer vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)4 m (13 ft)

A (εr 1.4 to 1.9)7 m (23 ft)

B (εr 1.9 to 4)

13 m (43 ft)

C (εr 4 to 10)

28 m (92 ft)

D (εr >10)44 m (144 ft)

Measurement in vessel with agitator
Vessel with agitator - measuring conditions
Turbulent medium surface (e.g. from filling from above, stirrers and baffles)

68

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Integrated antenna, PEEK, 20 mm (0.75 in) in vessel with agitator
Media group

Measuring range

A (εr 1.4 to 1.9)

1 m (3.3 ft)

B (εr 1.9 to 4)

1.5 m (5 ft)

C (εr 4 to 10)

3 m (10 ft)

D (εr >10)

5 m (16 ft)

Integrated antenna, PEEK, 40 mm (1.5 in) in vessel with agitator
Media group

Measuring range

A0 (εr 1.2 to 1.4)

1 m (3.3 ft)

A (εr 1.4 to 1.9)

1.5 m (5 ft)

B (εr 1.9 to 4)

3 m (10 ft)

C (εr 4 to 10)

7 m (23 ft)

D (εr >10)

11 m (36 ft)

Encapsulated antenna, PVDF, 40 mm (1.5 in) in vessel with agitator

Endress+Hauser

Media group

Measuring range

A0 (εr 1.2 to 1.4)

2 m (7 ft)

A (εr 1.4 to 1.9)

4 m (13 ft)

B (εr 1.9 to 4)

5 m (16.4 ft)

C (εr 4 to 10)

15 m (49.2 ft)

D (εr >10)20 m (65.6 ft)

69

Technical data

Micropilot FMR60B PROFIBUS PA

PTFE Drip-off antenna, 50 mm (2 in) in vessel with agitator

Operating frequency

Media group

Measuring range

A0 (εr 1.2 to 1.4)

2 m (7 ft)

A (εr 1.4 to 1.9)

4 m (13 ft)

B (εr 1.9 to 4)

7 m (23 ft)

C (εr 4 to 10)

15 m (49 ft)

D (εr >10)

25 m (82 ft)

Approx. 80 GHz
Up to 8 devices can be installed in a tank without the devices mutually influencing one
another.

Transmission power• Peak power: <1.5 mW• Average output power: <70 µW

15.2
PROFIBUS PA

OutputAccording to EN 50170 Volume 2, IEC 61158-2
Signal coding:
Manchester Bus Powered (MBP) type 1
Data transmission rate:
31.25 kBit/s, voltage mode
Galvanic isolation:
Yes

Signal on alarm

Diagnostics in accordance with PROFIBUS PA Profile 3.02
Local display
Status signal (in accordance with NAMUR Recommendation NE 107):
Plain text display
Operating tool via service interface (CDI)
Status signal (in accordance with NAMUR Recommendation NE 107):
Plain text display
Operating tool via PROFIBUS PA communication
Status signal (in accordance with NAMUR Recommendation NE 107):
Plain text display

Linearization

70

The linearization function of the device allows the conversion of the measured value into
any unit of length, weight, flow or volume.

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Pre-programmed linearization curves
Linearization tables for calculating the volume in the following vessels are preprogrammed
into the device:• Pyramid bottom• Conical bottom• Angled bottom• Horizontal cylinder• Sphere
Other linearization tables of up to 32 value pairs can be entered manually.
Protocol-specific data

PROFIBUS PA
Manufacturer ID:
17 (0x11)
Ident number:
0x1568 or 0x9700
Profile version:
3.02
GSD file and version
Information and files available at:• www.endress.com
On the product page for the device: Documents/Software → Device drivers• www.profibus.com
Output values
Analog Input:• Level linearized• Distance• Volume• Terminal voltage• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index, optional (Guidance → Heartbeat Technology → Build-up detection→ Build-up index)• Foam index, optional (Diagnostics → Heartbeat Technology → Foam detection → Foam
index)
Digital Input:
Is only available if the "Heartbeat Verification + Monitoring" application package wasselected.
• 168 Build-up detected, optional (Guidance → Heartbeat Technology → Build-up
detection → 168 Build-up detected)• 952 Foam detected, optional (Guidance → Heartbeat Technology → Foam detection →952 Foam detected)
Input values
Analog Output:
Analog value from PLC to be indicated on the display

Endress+Hauser

71Technical data

Micropilot FMR60B PROFIBUS PA

Supported functions• Identification & maintenance
Straightforward device identification on the part of the control system and nameplate• Automatic Ident Number adoption
GSD compatibility mode for generic profile 0x9700 "Transmitter with 1 Analog Input"• Physical layer diagnostics
Installation check of the PROFIBUS segment and device using terminal voltage andmessage monitoring• PROFIBUS upload/download
Reading and writing parameters is up to ten times faster with PROFIBUS upload/
download• Condensed status
Straightforward and self-explanatory diagnostic information by categorizing diagnostic
messages that occur

15.3
Ambient temperature
range

Environment

The following values apply up to a process temperature of +85 °C (+185 °F). At higher
process temperatures, the permitted ambient temperature is reduced.
• Without LCD display:Standard: –40 to +85 °C (–40 to +185 °F)• With LCD display: –40 to +85 °C (–40 to +185 °F) with limitations in optical properties
such as display speed and contrast for example. Can be used without limitations up to–20 to +60 °C (–4 to +140 °F)
If operating outdoors in strong sunlight:• Mount the device in the shade.
• Avoid direct sunlight, particularly in warm climatic regions.
• Use a weather protection cover (see accessories).

Ambient temperature limits

The permitted ambient temperature (Ta) depends on the selected housing material(Product Configurator → Housing; Material →) and the selected process temperature range(Product Configurator → Application →).
In the event of temperature (Tp) at the process connection, the permitted ambient
temperature (Ta) is reduced.
The following information only takes functional aspects into consideration. Additional
restrictions may apply for certified device versions.

72

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Plastic housing
Plastic housing; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 24
P1
P2
P3
P4
P5

Plastic housing; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –20 to +150 °C (–4 to +302 °F) is limited to
0 to +150 °C (+32 to +302 °F).
Restriction to a process temperature of 0 to +150 °C (+32 to +302 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 25
P1
P2
P3
P4
P5

Endress+Hauser

Plastic housing; process temperature 0 to +150 °C (+32 to +302 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

73

Technical data

Micropilot FMR60B PROFIBUS PA

Plastic housing; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 26
P1
P2
P3
P4
P5

Plastic housing; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –20 to +200 °C (–4 to +392 °F) is limited to
0 to +200 °C (+32 to +392 °F).
Restriction to a process temperature of 0 to +200 °C (+32 to +392 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 27
P1
P2
P3
P4
P5

74

Plastic housing; process temperature 0 to +200 °C (+32 to +392 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Plastic housing; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 28
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +80 °C (+176 °F) | Ta: +75 °C (+167 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of–40 to +80 °C (–40 to +176 °F) is limited to 0 to +80 °C (+32 to +176 °F).
Restriction to a process temperature of 0 to +80 °C (+32 to +176 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 29
P1
P2
P3
P4
P5

Endress+Hauser

Plastic housing; process temperature 0 to +80 °C (+32 to +176 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +80 °C (+176 °F) | Ta: +75 °C (+167 °F)= Tp: +80 °C (+176 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

75

Technical data

Micropilot FMR60B PROFIBUS PA

Plastic housing; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 30
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +130 °C (+266 °F) | Ta: +41 °C (+106 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –40 to +130 °C (–40 to +266 °F) is limited to
0 to +130 °C (+32 to +266 °F).
Restriction to a process temperature of 0 to +130 °C (+32 to +266 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 31
P1
P2
P3
P4
P5

Plastic housing; process temperature 0 to +130 °C (+32 to +266 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +130 °C (+266 °F) | Ta: +41 °C (+106 °F)= Tp: +130 °C (+266 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Plastic housing; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 32
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –40 to +150 °C (–40 to +302 °F) is limited to
0 to +150 °C (+32 to +302 °F).
76

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Restriction to a process temperature of 0 to +150 °C (+32 to +302 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 33
P1
P2
P3
P4
P5

Plastic housing; process temperature 0 to +150 °C (+32 to +302 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Plastic housing; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 34
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –40 to +200 °C (–40 to +392 °F) is limited to
0 to +200 °C (+32 to +392 °F).
Restriction to a process temperature of 0 to +200 °C (+32 to +392 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 35
P1
P2
P3
P4
P5

Endress+Hauser

Plastic housing; process temperature 0 to +200 °C (+32 to +392 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

77

Technical data

Micropilot FMR60B PROFIBUS PA

Aluminum housing, coated
Aluminum housing; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 36
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +150 °C (+302 °F) | Ta: +53 °C (+127 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

Aluminum housing; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 37
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +200 °C (+392 °F) | Ta: +47 °C (+117 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

Aluminum housing; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 38
P1
P2
P3
P4
P5

78

Aluminum housing, coated; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +80 °C (+176 °F) | Ta: +79 °C (+174 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Aluminum housing; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 39
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +130 °C (+266 °F) | Ta: +55 °C (+131 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Aluminum housing; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 40
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +150 °C (+302 °F) | Ta: +53 °C (+127 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Aluminum housing; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 41
P1
P2
P3
P4
P5

Endress+Hauser

Aluminum housing, coated; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +200 °C (+392 °F) | Ta: +47 °C (+117 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

79Technical data

Micropilot FMR60B PROFIBUS PA

316L housing
316L housing; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 42
P1
P2
P3
P4
P5

316L housing; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +150 °C (+302 °F) | Ta: +43 °C (+109 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 43
P1
P2
P3
P4
P5

316L housing; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +200 °C (+392 °F) | Ta: +38 °C (+100 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 44
P1
P2
P3
P4
P5

80

316L housing; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +80 °C (+176 °F) | Ta: +77 °C (+171 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

316L housing; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 45
P1
P2
P3
P4
P5

316L housing; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +130 °C (+266 °F) | Ta: +54 °C (+129 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 46
P1
P2
P3
P4
P5

316L housing; process temperature range: –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +150 °C (+302 °F) | Ta: +43 °C (+109 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 47
P1
P2
P3
P4
P5

Endress+Hauser

316L housing; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +200 °C (+392 °F) | Ta: +38 °C (+100 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

81

Technical data

Micropilot FMR60B PROFIBUS PA

316L housing, hygiene
316L housing, hygiene; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 48
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +41 °C (+106 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing, hygiene; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 49
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +32 °C (+90 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing, hygiene; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 50
P1
P2
P3
P4
P5

82

316L housing, hygiene; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +80 °C (+176 °F) | Ta: +75 °C (+167 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

316L housing, hygiene; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 51
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +130 °C (+266 °F) | Ta: +55 °C (+131 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing, hygiene; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 52
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature range: –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +41 °C (+106 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing, hygiene; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 53
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +32 °C (+90 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Storage temperature• Without LCD display: –40 to +90 °C (–40 to +194 °F)• With LCD display: –40 to +85 °C (–40 to +185 °F)

Climate class

DIN EN 60068-2-38 (test Z/AD)

Endress+Hauser

83Technical data

Micropilot FMR60B PROFIBUS PA

Installation height as per
IEC61010-1 Ed.3

Generally up to 5 000 m (16 404 ft) above sea level

Degree of protection

Test as per IEC 60529 and NEMA 250-2014
Housing
IP66/68, NEMA TYPE 4X/6P
IP68 test condition: 1.83 m under water for 24 hours.
Cable entries• Gland M20, plastic, IP66/68 NEMA TYPE 4X/6P• Gland M20, nickel-plated brass, IP66/68 NEMA TYPE 4X/6P• Gland M20, 316L, IP66/68 NEMA TYPE 4X/6P• Gland M20, hygiene, IP66/68/69 NEMA Type 4X/6P• Thread M20, IP66/68 NEMA TYPE 4X/6P• Thread G1/2, IP66/68 NEMA TYPE 4X/6P
If the G1/2 thread is selected, the device is delivered with an M20 thread as standard
and a G1/2 adapter is included with the delivery, along with the corresponding
documentation• Thread NPT1/2, IP66/68 NEMA TYPE 4X/6P• M12 plug
• When housing is closed and connecting cable is plugged in: IP66/67 NEMA TYPE 4X• When housing is open or connecting cable is not plugged in: IP20, NEMA TYPE 1
NOTICEM12 plug: Loss of IP protection class due to incorrect installation!
‣ The degree of protection only applies if the connecting cable used is plugged in andscrewed tight.
‣ The degree of protection only applies if the connecting cable used is specified according
to IP66/67 NEMA 4X.
‣ The protection classes are only maintained if the dummy cap is used or the cable isconnected.

Vibration resistanceDIN EN 60068-2-64 / IEC 60068-2-64 for 5 to 2 000 Hz: 1.5 (m/s2)2/Hz

Electromagnetic
compatibility (EMC)• Electromagnetic compatibility as per EN 61326 series and NAMUR recommendation
EMC (NE21)• Maximum measured error during EMC testing: < 0.5 % of the current digital measured
value
For more details refer to the EU Declaration of Conformity.

84

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

15.4
Process pressure range

Process

LWARNING
The maximum pressure for the device depends on the lowest-rated component withregard to pressure (components are: process connection, optional mounted parts oraccessories).
‣ Only operate the device within the specified limits for the components!
‣ MWP (Maximum Working Pressure): The MWP is specified on the nameplate. Thisvalue refers to a reference temperature of +20 °C (+68 °F) and may be applied to thedevice for an unlimited time. Note temperature dependence of MWP. For flanges, referto the following standards for the permitted pressure values at higher temperatures:
EN 1092-1 (with regard to their stability/temperature property, the materials 1.4435
and 1.4404 are grouped together under EN 1092-1; the chemical composition of thetwo materials can be identical), ASME B16.5, JIS B2220 (the latest version of thestandard applies in each case). MWP data that deviate from this are provided in therelevant sections of the Technical Information.
‣ The Pressure Equipment Directive (2014/68/EU) uses the abbreviation PS . This
corresponds to the maximum working pressure (MWP) of the device.
The following tables show the dependencies between the seal material, processtemperature (TP) and process pressure range for each process connection that can beselected for the antenna used.
Encapsulated antenna, PVDF, 40 mm (1.5 in)Process connection thread 1-½"
Seal

Tp

Process pressure range

PVDF encapsulated–40 to +80 °C(–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

PVDF encapsulated–40 to +130 °C(–40 to +266 °F)–1 to 3 bar (–14.5 to 43.5 psi)

The following temperature restriction applies for devices with the dust ignition-proof
approval category 1D, 2D or 3D
A0047831

PVDF encapsulated–20 to +80 °C (–4 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

Seal

Tp

Process pressure range

PVDF encapsulated–40 to +80 °C(–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

Process connection UNI flange PP

The following temperature restriction applies for devices with the dust ignition-proof
approval category 1D, 2D or 3D
PVDF encapsulated–20 to +80 °C (–4 to +176 °F) –1 to 3 bar (–14.5 to 43.5 psi)

A0047947

The pressure range may be further restricted in the event of a CRN approval.

Endress+Hauser

85

Technical data

Micropilot FMR60B PROFIBUS PA

Drip-off antenna 50 mm (2 in)Process connection thread

A0047447

Seal

Tp

Process pressure range

FKM Viton GLT–40 to +130 °C (–40 to +266 °F)–1 to 16 bar (–14.5 to 232 psi)

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 16 bar (–14.5 to 232 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 16 bar (–14.5 to 232 psi)

EPDM

–40 to +130 °C (–40 to +266 °F)–1 to 16 bar (–14.5 to 232 psi)

HNBR

–20 to +150 °C (–4 to +302 °F)–1 to 16 bar (–14.5 to 232 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 16 bar (–14.5 to 232 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 16 bar (–14.5 to 232 psi)

Process connection UNI flange PP
Seal

Tp

Process pressure range

FKM Viton GLT–40 to +80 °C (–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

EPDM

–40 to +80 °C (–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

The following temperature restriction applies for devices with the HNBR or FFKM
Kalrez O-ring
A0047726

HNBR–20 to +80 °C (–4 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FFKM Kalrez–20 to +80 °C (–4 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

Process connection UNI flange 316L

A0047726

Seal

Tp

Process pressure range

FKM Viton GLT–40 to +130 °C (–40 to +266 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 3 bar (–14.5 to 43.5 psi)

EPDM

–40 to +130 °C (–40 to +266 °F)–1 to 3 bar (–14.5 to 43.5 psi)

HNBR

–20 to +150 °C (–4 to +302 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 3 bar (–14.5 to 43.5 psi)

The pressure range may be further restricted in the event of a CRN approval.
Integrated antenna, PEEK, 20 mm (0.75 in)Process connection thread 3/4"
Seal

Tp

Process pressure range

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

A0047832

The pressure range may be further restricted in the event of a CRN approval.

86

Endress+Hauser

Micropilot FMR60B PROFIBUS PA

Technical data

Integrated antenna, PEEK, 40 mm (1.5 in)Process connection thread 1-½"
Seal

Tp

Process pressure range

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

A0047833

The pressure range may be further restricted in the event of a CRN approval.

Dielectric constant

For liquidsεr ≥ 1.2
Contact Endress+Hauser for applications with lower dielectric constants than indicated.

Endress+Hauser87IndexMicropilot FMR60B PROFIBUS PA

Index
A

O

B

Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Purpose of this document . . . . . . . . . . . . . . . . . . . . . . 5

Access authorization to parameters
Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Access code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Incorrect input . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Bluetooth® wireless technology . . . . . . . . . . . . . . . . . 31
Bus address . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30

C

CE mark (Declaration of Conformity) . . . . . . . . . . . . . . 9
Cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Climate class . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 83

D

Declaration of Conformity . . . . . . . . . . . . . . . . . . . . . . 9
Device locking, status . . . . . . . . . . . . . . . . . . . . . . . . 44
Device master file . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
Device Viewer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Diagnostic event . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
DIP switches . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Display values
For locking status . . . . . . . . . . . . . . . . . . . . . . . . . 44
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Document
Purpose . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

E

EMPTY_MODULE . . . . . . . . . . . . . . . . . . . . . . . . . . . 38
Event history . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Event text . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55

Operating keys . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29
Operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 44
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Output values . . . . . . . . . . . . . . . . . . . . . . . . . . . 34, 71

P

R

Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Reading off measured values . . . . . . . . . . . . . . . . . . . 44
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . . . 7
Repair concept . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
Requirements concerning the staff . . . . . . . . . . . . . . . . 8
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57

S

Safety instructions
Basic . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Safety Instructions (XA) . . . . . . . . . . . . . . . . . . . . . . . . 6
Securing screw . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22
Service interface (CDI) . . . . . . . . . . . . . . . . . . . . . . . . 41
Settings
Adapting the measuring device to the process
conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 44
Software addressing . . . . . . . . . . . . . . . . . . . . . . . . . 30
Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
Nameplate . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
Status signal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Submenu
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51

T

Troubleshooting . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Turning the display module . . . . . . . . . . . . . . . . . . . . 18

F

U

G

W

Field of application
Residual risks . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Filtering the event logbook . . . . . . . . . . . . . . . . . . . . 52
GSD . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35

H

Use of the measuring device
see Intended use
Use of the measuring devices
Borderline cases . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Incorrect use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30

Hardware addressing . . . . . . . . . . . . . . . . . . . . . . . . . 29

I

Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

MMedia . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

88Endress+Hauser

*71627296*
71627296

www.addresses.endress.com