Installation Instruction

Sensor FMD71, FMD72 and cables
Deltabar FMD71, FMD72
EA00066P/00/A2/03.13
71235704
71235704

Instruction is valid for the following spare part sets:
Order number

Device component

Connection of

XPP0008-#

Sensor FMD71

Deltabar FMD71

XPP0007-#

Sensor FMD72

Deltabar FMD72

71186132

Connection cable L=6 ft; 1.82 m

Sensor - Sensor, Sensor- Transmitter

Deltabar FMD71, FMD72

71186136

Connection cable L=15 ft; 4.57 m

Sensor - Sensor, Sensor - Transmitter

Deltabar FMD71, FMD72

71186138

Connection cable L=35 ft; 10.67 m

Sensor - Sensor, Sensor - Transmitter

Deltabar FMD71, FMD72

71186141

Connection cable L=100 ft; 30.48 m

Sensor - Sensor

Deltabar FMD71, FMD72

71243224

Connection cable L=150 ft; 45.72 m

Sensor - Sensor

Deltabar FMD71, FMD72

We recommend that the Installation Instructions be kept with the packaging at all times.

Usage

Sensor FMD71, FMD72 and cables

Confirmation whether the spare part is permitted to be used with the measuring device
The spare parts set and Installation Instructions are used to replace a faulty unit with a functioning unit of the same type. Use genuine parts from Endress+Hauser only.
Only original spare parts supplied by Endress+Hauser shall be used with the measuring device. Therefore, before use, check whether the sparepart set is compatible with the measuring device.
A spare parts overview label is located in the connection compartment cover of the measuring device. If there is no label or the spare part is notlisted the appropriate spare part can also be identified via W@M Device Viewer.
How carrying out such a check via the W@M Device Viewer is described below:

Device Viewer
Ser.No.:1234567890
Ser.
No.:12345

Ser. No.:12345

ii

www.endress.com/deviceviewer=?

Choose the Endress+Hauser Device Viewer via web browser: www.endress.com/deviceviewer
Enter the serial number (Ser. No.) of the device into the W@M Device Viewer (on the label of the transmitter), then click on "Spare parts".
The list of the available spare parts for the device is displayed.
Check the order number on the packaging label of the spare part set.
Check whether the order number of the spare part set is specified in the list displayed by the device viewer:= YES, the spare part can be used.
= NO, the spare part may not be used.

Overview of the personnel authorized to carry out repairs
Authorization to carry out a repair depends on the approval of the measuring device. The table shows the respective group of persons for each.

!

Note!
The person who carries out the repair is responsible for safety during the work, the quality of work completed and safety of the device after repair.
Approval of the measuring device

Group of persons authorized to carry out repairs

without approval1, 2, 3

with approval (for Ex. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 Trained customer technician
2 Service technician authorized by Endress+Hauser
3 Endress+Hauser (send measuring device back to manufacturer)

2

Endress + Hauser

Sensor FMD71, FMD72 and cables

Safety instructions• Check whether the spare part matches the identification label on themeasuring device, as explained on the first page.
• The spare parts set and Installation Instructions are used to replace a faulty unit with a functioning unit of the same type. Use genuine parts from Endress+Hauser only.
• In the case of Ex-certified measuring devices: Observe theinformation in the documentation "Safety Instructions" (XA, ZD).
The relevant Safety Instructions (XA) are indicated on the nameplate.
• The measuring device is energized. Danger: Risk of electric shock!
Open the measuring device in a de-energized state only.
• Before removing the device: set the process in a safe condition andpurge the pipe of dangerous materials.
• Hot surfaces! Risk of injury! Before commencing work, allow thesystem and measuring device to cool down to a touchable
temperature.
• In the case of measuring devices in custody transfer, the custodytransfer status no longer applies once the lead seal has been removed.
• Comply with national regulations governing mounting, electricalinstallation, commissioning, maintenance and repair procedures.
• Requirements with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance andrepair of the measuring devices:– trained in instrument safety– familiar with the individual operation conditions of the devices– for Ex-certified measuring devices: also trained in explosion
protection• Follow the Operating Instructions for the device.

Endress + Hauser• Risk of damaging electronic components! Ensure you have a working
environment protected from electrostatic discharge.
• After removing the electronics cover, there is a risk of electric shock
as shock protection is removed! Switch off the measuring device before removing internal covers.
• Modifications to the measuring device are not permitted.
• In the case of measuring devices in safety-related applications inaccordance with IEC 61508 or IEC 61511: After repair recommissionin accordance with Operating Instructions. Document the repairprocedure.
• Only open housing for a brief period. Avoid the penetration of foreign
bodies, moisture or contaminants.
• Replace defective seal/gaskets with genuine parts fromEndress+Hauser only.
• If threads are damaged or defective, the measuring device must berepaired.
• If spacing is reduced or the dielectric strength of the measuring device
cannot be guaranteed during repair work, perform a test on
completion of the work (e.g. high-voltage test in accordance with the manufacturer'sinstructions).
• Service connector:– do not connect in potentially explosive atmospheres.
– only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device
outlined in the Operating Instructions.
• If you have any questions, contact your E+H service organization.

3

Sensor FMD71, FMD72 and cables

Tool list

Size 1, 220, 24, 26 mm

Using a sensor FMD71, FMD72 as sensor HP or sensor LP
Each sensor can be used as sensor HP or sensor LP. With commissioning
we recommend to add the designation HP or LP on the the type plate.

Sensor module Deltabar FMD72
Ser. no.: MMMM6VY0335
P 0... 4000inH2O

MWP 400 mbar
CSA 041509834X
0F10524.5C

Non-Incendive forCl.I Div.2 Groups ABCD
Dual seal

IP66/IP68
TYPEXXX ENCL

HP (bottom)

HP (bottom)

LP (top)

only to be used withinMat.: 1.4401/316L ECTFE>316L a FMD72 system

Sticker

The sensor is prepared for the use as Sensor HP ex works. For use as Sensor LP, please observe the following:
123424 mm
Sensor3Sensor424 mm
Sensor5Sensor6

Exchange of sensor FMD71, FMD72• The designation LP/HP indicates, where the sensor is typicallyinstalled.

• For flange mounting, the seal is not allowed to press against theprocess isolating diaphragm.

Sensor
LPFilterSensor
HPTransmitter

Sensor1• Mount the sensor with the pressure compensation pointing
downwards.
• Keep the pressure compensation and GORE-TEX® filter free fromcontamination.
• Do not clean or touch process isolating diaphragms with hard or
pointed objects.

Sensor2• Due to the orientation of the sensor, there may be a shift in the zero
point, i.e. when the container is empty or partially full, the measured
value does not display zero. You can correct this zero point shift, this 
prodedure is described in the operating manual.

If you change both sensors, it is necessary to assign, which sensor is connected to the transmitter. This defines the corresponding sensor as the master. After commissioning, the second sensor is detected as the slave.
Menu path: Expert -> System -> Management -> Connect transm. (Direct Access "286").

4Endress + Hauser

Sensor FMD71, FMD72 and cables

Connection of Sensor LP with Sensor HPA Sensor LP (M20 / NPT 1/2)
123420 mm

1xNPT1/2

24 mm

M20Cable1

Cable2

Cable3

Cable4

B Sensor HP (M20 / NPT1/2)
1a1b24 mm

24 mm

26 mm
Cable5

Cable6

C Electrical connection
1 2 3 41243

1 = BK
2 = BU
3 = WH
4 = BN

Connect1!

Note!
The screening with cable shield is described in the associateddocumentation SD00354P.

Endress + Hauser

5Sensor FMD71, FMD72 and cables

Connection of sensor HP with transmitter#Warning!
Do not use the connection cables 100 ft; 30.48 m and 150 ft; 45.72 m to connect the sensor HP with the transmitter! A Sensor HP (M20 / NPT 1/2)
123420 mm

NPT1/21xM2024 mm
Cable7

Cable8

Cable9Cable10

B Transmitter (M20 / NPT1/2)
1a1b24 mm

24 mm
26 mm

Cable12

Cable11

C Electrical connection

124

3

1 = BK
2 = BU
3 = WH
4 = BN

1 2 3 4

Connect2!

Note!
The screening with cable shield is described in the associateddocumentation SD00354P.

6Endress + Hauser

Analyse

Einbauanleitung

Sensor FMD71, FMD72 und Kabel
Deltabar FMD71, FMD72
EA00066P/00/A2/03.1371235704

Die Einbauanleitung ist für folgende Ersatzteilsets gültig:
Bestellnummer

Original-Ersatzteilset

Verbindung von

XPP0008-########Sensor FMD71

Deltabar FMD71

XPP0007-########

Sensor FMD72

Deltabar FMD72

71186132

Verbindungskabel L=6 ft; 1,82 m

Sensor - Sensor, Sensor- Transmitter

Deltabar FMD71, FMD72

71186136

Verbindungskabel L=15 ft; 4,57 m

Sensor - Sensor, Sensor - Transmitter

Deltabar FMD71, FMD72

71186138

Verbindungskabel L=35 ft; 10,67 m

Sensor - Sensor, Sensor - TransmitterDeltabar FMD71, FMD72

71186141

Verbindungskabel L=100 ft; 30,48 m

Sensor - Sensor

Deltabar FMD71, FMD72

71243224

Verbindungskabel L=150ft; 45,72m

Sensor - Sensor

Deltabar FMD71, FMD72

Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

VerwendungSensor FMD71, FMD72 und KabelÜberprüfung, ob die Verwendung des Ersatzteils für das Messgerät erlaubt ist.
Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen eine funktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile von Endress+Hauser verwendet werden.
Grundsätzlich dürfen nur Ersatzteilsets verwendet werden, die von Endress+Hauser für das Messgerät vorgesehen sind. Vor der Verwendung ist
deshalb zu überprüfen, ob das Ersatzteilset zum Messgerät passt.
Im Anschlussraumdeckel des Messgeräts befindet sich ein Übersichtsschild Ersatzteile. Falls dieses nicht vorhanden oder das Ersatzteilset nicht
aufgelistet ist, kann eine solche Überprüfung via W@M Device Viewer durchgeführt werden.
Wie eine solche Überprüfung via W@M Device Viewer durchgeführt werden kann, ist nachfolgend beschrieben:

Device Viewer
Ser.No.:1234567890
Ser.
No.:12345

Ser. No.:12345

iiwww.endress.com/deviceviewer=?

Über einen Webbrowser den Endress+Hauser Device Viewer aufrufen: www.endress.com/deviceviewer
Die Seriennummer (Ser. No.) des Messgeräts (vom Typenschild des Messumformers) in den Device Viewer eingeben, danach auf den Reiter"Ersatzteile" klicken.
Auf dem Bildschirm wird eine Liste aller zur Verfügung stehenden Ersatzteilsets für das Messgerät angezeigt.
Die Bestellnummer des Ersatzteilsets ermitteln (auf dem Produktaufkleber der Verpackung).
Überprüfen, ob die Bestellnummer des Ersatzteilsets in der Liste der angezeigten Ersatzteile im Device Viewer vorhanden ist:
= JA, das Ersatzteilset darf für das Messgerät verwendet werden.
= NEIN, das Ersatzteilset darf für das Messgerät nicht verwendet werden.

Übersicht der reparaturberechtigten Personen
Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts abhängig. Die Tabelle zeigt den jeweils berechtigten
Personenkreis.

!

Hinweis!
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die Sicherheit während der Arbeiten, die Qualität der Ausführung
und die Sicherheit des Geräts nach der Reparatur.
Zulassung des Messgeräts

Reparaturberechtigter Personenkreis

ohne Zulassung

1, 2, 3

mit Zulassung (z.B. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 Ausgebildete Fachkraft des Kunden
2 Von Endress+Hauser autorisierter Servicetechniker
3 Endress+Hauser (Messgerät an Hersteller zurücksenden)

2Endress + Hauser

Sensor FMD71, FMD72 und Kabel

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf demMessgerät passt, wie auf der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit
gegen eine funktionierende Einheit des gleichen Typs zu ersetzen. Nur Originalteile von Endress+Hauser verwenden.
• Bei Ex-zertifizierten Messgeräten: Sicherheitshinweise im separaten
Dokument "Saftey Instructions" (XA, ZD) beachten.
Auf dem Typenschild ist angegeben, welche Sicherheitshinweise fürdas jeweilige Gerät relevant sind.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag.
Messgerät nur im spannungslosen Zustand öffnen.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen undLeitung von gefährlichen Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn:
Anlage und Messgerät auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen
der Plombe ist der geeichte Zustand aufgehoben.
• Nationale Vorschriften bezüglich der Montage, elektrischen
Installation, Inbetriebnahme, Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage,elektrische Installation, Inbetriebnahme, Wartung und Reparatur derMessgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten zusätzlich im Explosionsschutz
ausgebildet.

Endress + Hauser• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte
Arbeitsumgebung herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durchaufgehobenen Berührungsschutz! Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw. IEC 61511: Nach Reparatur Neuinbetriebnahme
gemäß Betriebsanleitung durchführen. Reparatur dokumentieren.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern,Feuchtigkeit oder Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen vonEndress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die
Spannungsfestigkeit des Messgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– nicht in explosionsfähiger Atmosphäre anschließen.
– nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport
und zur Rücksendung beachten.
• Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser
Serviceorganisation.

3Sensor FMD71, FMD72 und Kabel

Werkzeugliste

Size 1, 2

20, 24, 26 mm

Einsatz eines Sensors FMD71, FMD72 als Sensor LP oder
Sensor HP
Die Sensoren FMD71, FMD72 können sowohl als Sensor LP als auchSensor HP eingesetzt werden, deshalb sollte die Bezeichnung LP oderHP mit der Inbetriebnahme auf das Typenschild aufgeklebt werden.

Sensor module Deltabar FMD72
Ser. no.: MMMM6VY0335
P 0... 4000inH2O

MWP 400 mbar
CSA 041509834X
0F10524.5C

Non-Incendive forCl.I Div.2 Groups ABCD
Dual seal

IP66/IP68
TYPEXXX ENCL

HP (bottom)

HP (bottom)

LP (top)

only to be used withinMat.: 1.4401/316L ECTFE>316L a FMD72 system

Sticker

Ab Werk ist der Sensor FMD71, FDM 72 für den Einsatz als Sensor HP
vorbereitet. Für den Einsatz als Sensor LP ist folgendes zu beachten:
123424 mm

24 mmSensor3Sensor4

Sensor5

Sensor6

Austausch des Sensors FMD71, FMD72• Die Bezeichnungen LP und HP zeigen an, wo der Sensor typischerweise eingebaut wird.

• Bei der Flanschmontage darf die Dichtung nicht auf die Prozessmembrane drücken.

Sensor
LPFilter
SensorHP

Transmitter

Sensor1• Montieren Sie den Sensor so, dass der Druckausgleich möglichst nachunten zeigt.
• Druckausgleich und GORE-TEX® BEiFilter frei vonVerschmutzungen halten.
• Prozessmembrane nicht mit spitzen und harten Gegenständen
eindrücken oder reinigen.

Sensor2• Bedingt durch die Einbaulage der Sensormodule kann es zu einer
Nullpunktverschiebung kommen, d.h. bei leerem oder teilbefülltem
Behälter zeigt der Messwert nicht Null an. Diese Nullpunktverschiebung können Sie korrigieren, nehmen Siedazu die Bedienungsanleitung zur Hand.

Beim Austausch beider Sensoren muss in der Elektronik festgelegt werden, welcher Sensor (HP oder LP) mit dem Transmitter verbundenist. Dieser Sensor wird damit zum Master, der zweite Sensor wird nach Inbetriebnahme als Slave erkannt.
Die Einstellung erfolgt im Menüpfad: Experte -> System -> Verwaltung -> Transm.Anschl. (Direct Access "286").

4Endress + Hauser

Sensor FMD71, FMD72 und Kabel

Verbindung zwischen Sensor LP und Sensor HP
A Sensor LP (M20 / NPT1/2)
123420 mm

1xNPT1/2

24 mm

M20
Cable1

Cable2

Cable3

Cable4

B Sensor HP (M20 / NPT 1/2)
1a1b24 mm

24 mm

26 mm
Cable5

Cable6

C Elektrischer Anschluss
11 2 3 41243

1 = BK
2 = BU
3 = WH
4 = BN

Connect1! Hinweis!
Die Erdung mit Kabelschirmung ist in der mitgelieferten
Dokumentation SD00354P beschrieben.

Endress + Hauser

5Sensor FMD71, FMD72 und Kabel

Verbindung zwischen Sensor HP und Transmitter#Warnung!
Verbindungskabel 100 ft; 30,48 m und 150 ft; 42,72 m nicht für die Verbindung zwischen Senor HP und Transmitter verwenden!

A Sensor HP (M20 / NPT 1/2)
123420 mm1xNPT1/2
M2024 mmCable7

Cable8

Cable9Cable10

B Transmitter (M20 / NPT 1/2)
1a1b24 mm

24 mm
26 mm

Cable12

Cable11

C Elektrischer Anschluss1124

3

1 = BK
2 = BU
3 = WH
4 = BN

1 2 3 4

Connect2! Hinweis!
Die Erdung mit Kabelschirmung ist in der mitgelieferten
Dokumentation SD00354P beschrieben.

6Endress + Hauser