A TOPCON PRE-EXAM SOLUTION

Specifications
Refractive Power
Measurement

Corneal Curvature
Measurement

2013_ 12_24
No.0001
SN

Cylindrical refractive power

0D to ±10D (0.12D/0.25D steps)

Astigmatic axial angle

0°to 180° (in 1°or 5°steps)

Minimal measurable pupil diameter

φ2 mmCorneal curvature radius

5.00 to 10.00mm (0.01mm step)

Corneal refractive power

67.50D to 33.75D (0.12D/0.25D steps)

S

0D to ±10D (0.12D/0.25 D steps)

Corneal astigmatic axial angle

0°to 180° ( 1°/ 5°steps)

Range of Subjective

pherical refractive power:

Spherical refractive power: -18D to +18D (0.25D steps)

refractive check

Test chart:

VA

Eyesight test chart of 0.1 to 1.2 or 20/200 to 20/15, Grid display

Chart display:A

Overall, Horizontal series, Contrast change

VA

Test items:

2

NEAR TEST(REF)
DIST.
R

ADD

VA

L

ADD

VA

DIST.

GRID CHART(REF)

3

Printout Sample
4

Subjective refraction Far VA value

2

Subjective refraction Near VA value

3

Grid test result

56

Glare test result

5Contrast test result

6

Far VA for lensmeter

7

Near VA for lensmeter

8

Glare test VA for lensmeter

9

Contrast test VA for lensmeter

7L
NS:

NS:

TS:

C:
TI:

14

R
TS:

C:
NI:

Auto Kerato-Refractometer(where, corneal refractive power =1.3375)

AC

NI:

Far-sightedness, Near-sightedness, Glare test

PD Measurement Range

20mm to 85mm (0.5mm step)

Data Transport Terminal

USB (Import) /RS-232C (Import/Export) / LAN (Export)

Dimensions

317-341mm (W) × 521-538mm (D) × 447-477 mm (H)Weight15 kg

Power Supply

100-240V AC, 50-60Hz, 70VA

TI:※-25D ≦ spherical refractive power + cylindrical refractive power or spherical refractive power + cylindrical refractive power ≦ +22D

GLARE TEST(REF)
R
L
VA
VA
CONTRAST TEST(REF)
R

VA

LVL.

L

System chart
VA

LVL.

A

VA

C

A

VA

NEAR TEST(CL)
DIST.
R

ADD

VA

L

ADD

VA

SBJ. DATA(CL)
S
R
S

L

DIST.

Computerized
Lensmeter

Auto KeratoRefractometer

Power Supply Unit
w/Built in PC

KR-800S

Tablet

9R
L
VA
VA
CONTRAST TEST(CL)
R
VA
LVL.

Compu Vision

0.1
0.2

E

0.3

20/60

0.4

20/50
20/40

0.5
0.6

20/30

0.7

20/25

0.8

20/20

1.0

20/151.28

HBDV7

PHCT2

PVAD3

VSHE4

0.1
20/2000.2
20/100

0.3
20/80

0.4
20/30

0.5
20/25

LCD Chart(wire or wireless)Mirror Chart

Mouse & Monitor

LVA
LVL.

Chart thumbnail

20/80

Wave-Front
Analyzer

GLARE TEST(CL)8

Auto Chart
Projector
Power Supply Unit
w/External PC

Sample20/100Ez Meter

1 Dial Controller

TOPCON

20/200KR-800S※

Corneal astigmatic refractive power
AM 10 : 00
01

SBJ. DATA(REF)
SRL-25D to +22D (0.12D/0.25D steps) ※-KR 010001-

OID :NAME

1Spherical refractive power

PCMouse & Monitor* Please contact Topcon subsidiaries or dealers for system configurations.

* Not available in all countries, please check with your distributor for availability in your country* Subject to change in design and/or specifications without advanced notice.

IMPORTANTIn order to obtain the best results with this instrument,
please be sure to review all user instructions prior to operation.

Objective chart Subjective chart75-1 Hasunuma-cho, Itabashi-ku, Tokyo 174-8580, Japan. Phone:+81-(0)3-3558-2522/2502 Fax:+81-(0)3-3965-6898 www.topcon.co.jp

RDZT7ARFS6CNDT4KOZF5HKNS90.6
20/20

0.7
20/60

0.8
20/50

1.020/40

1.220/15

TOPCON MEDICAL SYSTEMS, INC.

TOPCON S.A.R.L.

TOPCON CANADA INC.

TOPCON DEUTSCHLAND MEDICAL G.m.b.H.

TOPCON EUROPE MEDICAL B.V.

TOPCON SCANDINAVIA A.B.

111 Bauer Drive, Oakland, NJ 07436, U.S.A.
Phone:+1-201-599-5100 Fax:+1-201-599-5250 www.topconmedical.com
110 Provencher Avenue, Boisbriand, QC J7G 1N1 CANADA
Phone:+1-450-430-7771 Fax:+1-450-430-6457 www.topcon.ca

Grid chart

Essebaan 11; 2908 LJ Capelle a/d IJssel; P.O.Box145;
2900 AC Capelle a/d IJssel; THE NETHERLANDS
Phone:+31 -(0)10-4585077 Fax:+31 -(0)10-4585045
E-mail: medical@topcon.eu; www.topcon-medical.eu

ITALY OFFICE

Viale dell' Industria 60; 20037 Paderno Dugnano; (Milano), ITALY
Phone:+39-02-9186671 Fax:+39-02-91081091 E-mail: info@topcon.it; www.topcon.it

DANMARK OFFICE

Praestemarksvej 25; 4000 Roskilde, DANMARK
Phone:+45-46-327500 Fax:+45-46-327555
E-mail: topcon@topcondanmark.dk www.topcondanmark.dk

IRELAND OFFICE

Unit 276, Blanchardstown; Corporate Park 2 Ballycoolin Dublin 15, IRELAND
Phone:+353-18975900 Fax:+353-18293915 E-mail: medical@topcon.ie; www.topcon.ie

BAT A1 3 route de la révolte 93206 SAINT DENIS CEDEX
Phone:+33-1-49-21-23-23 Fax : +33-1-49-21-23-24 E-mail:topcon@topcon.fr; www.topcon.fr
Hanns-Martin-Schleyer Strasse 41; D-47877 Willich, GERMANY
Phone:+49-(0)2154-8850 Fax:+49-(0)2154-885177 E-mail:med@topcon.de; www.topcon.deNeongatan 2; P.O.Box 25; 43151 Mölndal, SWEDEN
Phone:+46-(0)31-7109200 Fax:+46-(0)31-7109249 E-mail:medical@topcon.se; www.topcon.se

TOPCON ESPANA S.A.
HEAD OFFICE

Frederic Mompou 4 Esc. A Bajos 3, 08960 Sant Just Desvern Barcelona, SPAIN
Phone:+34-93-4734057 Fax:+34-93-4733932 E-mail: medica@topcon.es; www.topcon.es

TOPCON ( GREAT BRITAIN ) LTD.

Topcon House,Kennet Side,Bone Lane,Newbury,Berkshire RG14 5PX United Kingdom
Phone:+44-(0)1635-551120 Fax:+44-(0)1635-551170 E-mail:medical@topcon.co.uk; www.topcon.co.uk

TOPCON POLSKA Sp. z. o. o.

ul. Warszawska 23; 42-470 Siewierz, POLAND
Phone:+48-(0)32-6705045 Fax:+48-(0)32-6713405 www.topcon-polska.pl

TOPCON SINGAPORE MEDICAL PTE. LTD.

1 JALAN KILANG TIMOR #09-01 PACIFIC TECH CENTRE SINGAPORE 159303
Phone:+65-68720606 Fax:+65-67736150
E-mail:medical_sales@topcon.com.sg www.topcon.com.sg

REPRESENTATIVE OFFICE IN INDONESIA

Level 38, Tower A, Kota Kasablanka unit GH-04, JI.
Casablanca, Kav 88, Jakarta, Indonesia 12870
Phone:+62-21-2963-8004

TOPCON INSTRUMENTS ( MALAYSIA ) SDN.BHD.
No. D1, (Ground Floor), Jalan Excella 2, Off Jalan Ampang Putra,
Taman Ampang Hilir, 55100 Kuala Lumpur,MALAYSIA
Phone:+60-(0)3-42709866 Fax:+60-(0)3-42709766

TOPCON INSTRUMENTS ( THAILAND ) CO.,LTD.
77/162 Sinnsathorn Tower, 37th Floor,Krungthonburi Rd.,Klongtonsai,
Klongsarn, Bangkok 10600,THAILAND
Phone:+66(0)2-440-1152~7 Fax:+66-(0)2-440-1158

2016
Printed in Japan 2016_03_30_NP_KR800S_E148-1 GE

PERFORMANCEYOU CAN COUNT ON

KR-800S

More Than Just an Auto Kerato Refractometer
The sleek and modern design of the KR-800S incorporates the latest
technology designed to increase practice efficiency and enhance the patient
experience. The KR-800S provides auto refraction, keratometry, contrast
screening, visual acuity, and glare testing all in one easy to use instrument.

Auto Kerato-Refractometer

KR-800S will dramatically change
the clinicalcliniccal workfloww
Subjective
VA TestSubjective
IOL Type
Selection

Grid Test

Glare Test

Contrast
Test

Objective
Test

This multi-functionality supports enhanced work flow, and enables theKR-800S to be used in any variety of settings from hospitals, multi-specialty
ophthalmic and optometric clinics and even vision screening centers.

Enhances workflow efficiency and improves utilization.
Comfortable and smooth examination provides
superior patient experience
Assessment of visual acuity under glare conditions
Multi-functional instrument helps you quickly
document the patient's vision during treatment
and follow up of a range of ocular diseases
Small footprint and LAN connection supports
flexible installation.

Range of functions
Objective and Subjective Data

Glare test

KR-800S displays the test results of all objective and subjective measurements on a single screen. This all-in-one instrument

The test is simple, standardized and provides asaves examination time and offers wide functionality in situations where a manual refractor is not available, such as school

consistent and reliable way to evaluate visionscreening, athletic or occupational screening, and nursing homes. Refractive results can be viewed on the screen, printed out,changes in the presence of bright lights.

or exported to an Electronic Health Record.

*The Glare test can only be performed with the subjective far distance test.

Normal Vision

Glare VisionObjective Test

ExampleExampleContrast test
The contrast test is an ideal test to check thepatient’s quality of vision. The contrast of thechart can be changed across a range ofpercentages.
* The Contrast test can only be performed with the subjective far distance test.
* Only the contrast of the background changes.

Fixation target

OBJ/SBJ

Up-and-down auto tracking
FOG

Cataract

All clear

100 %Auto ShootGrid test
The grid test can be performed for conditions affecting the macula.
Patients with macular disease may see wavy or missing lines. The test canbe helpful in detecting early signs of abnormality in the eye.
New grid test function shows the grid for no longer than 0.25 seconds toSubjective Test

prevent the natural “completion phenomenon” filling in the grid.

All information isshown on the one screen
Lensmeter results, Objective results,Subjectiveresults,andSpherical

Equivalent simulation are displayed.

What is Completion phenomenon?

The operator can refer to all of theinformation during the subjectivetests.

The human brain is able to restore “crooked” lines
back to straight lines in just over 0.25 seconds,losing the actual result*The Grid test can only be performed with the subjective far distance test.

Far /Near
VA check/Grid checkContrast Test
Glare Test

Display all data

50 %

25 %

12.5 %

Other Features

Topcon’s Cataract Workstation
Cataract surgery quality control

VA Simulation Premium IOL

Visual acuity (VA) is the most common clinical measure

KR-800S offers a Spherical Equivalent mode which can simulate theThis software function allows the patient too visualize

following cataract surgery. It is how we describe and measure

benefit of a premium (toric) IOL.

etween a
the potential post-surgical difference between

the success of surgery. Measurement of VA must be

spherical and toric IOL. Simply push a single button to

standardized and systematic. Topcon’s KR-800S Autoswitch “Sphere, Cylinder, and Axis” to “Spherical

Kerato-refractometer with subjective VA check will doparison ofEquivalent”. The patient may then view a comparison

exactly that. With the KR-800S the VA can be subjectively

their with and without cylinder correction.

tested pre- and post-operative cataract surgery. With theThis feature is also useful for contact lenss patientsunique features of the KR-800S, such as “Glare” test andThe KR-800S completes the screening workflow of cataract surgery. Alluivalent tooconsidering a toric lens versus a spherical equivalent“Contrast” test, you can even evaluate the progression ofnecessary cataract pre-op information can be obtained by combining thecorrect low astigmatism.

cataract and identify cataract that is causing impaired quality

KR-800S and ALADDIN, while the KR-800S assists you postop in Visualof vision without a significant reduction in high contrast VA.

Acuity evaluation and determining the success of the cataract surgery.

Compare toric to
non toric correction

Toric IOL

SubjectiveIOL TypeSelection

ALADDIN and KR-800S - the perfect combination for your cataract

KR-800S

This allows the patient to visualize
the potential post-surgical differences
between a standard and Toric IOL

Cataract workstation

practice.

Auto Kerato-Refractometer

ALADDIN

Optical Biometry & Topography System

KR-800S

Rotary Prism Measuring System
With Topcon's exclusive Rotary Prism Technology, theAuto Kerato-Refractometer

Rotary Prism Measuring System (Image)

KR-800S also provides unmatched accuracy and reliability.
The innovative optical design incorporated into theKR-1W

Rotary Prism

Pre-Operative

KR-800S allows for accurate and reliable measurements
wit h a pupil as small as φ2mm.
MirrorRotary Ring Image

Measuring
RingSensor

Wave-Front Analyzer

Subjective Refraction
and Pre-OP-diagnostics
Pupillography
Topography
Biometry inkl. K1 & K2
IOL Calculation

Cataract
Surgery
Post-Operative
Subjective Refraction
and Post-OP-diagnostics

Follow up