BA02266F/00/EN/01.23-00
71610086
2023-04-22

Products

Solutions

01.00.zz (Device firmware)

Operating Instructions
Micropilot FMR60B
PROFINET with Ethernet-APL
Free space radar

advanced physical layer

Services

Micropilot FMR60B PROFINET with Ethernet-APL

1.

Order code: XXXXX-XXXXXX
Ser. no.:XXXXXXXXXXXX
Ext. ord. cd.: XXX.XXXX.XX

Serial number

2. www.endress.com/deviceviewer

Endress+Hauser
Operations App

3.

A0023555• Make sure the document is stored in a safe place such that it is always available whenworking on or with the device• Avoid danger to individuals or the facility: read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures
The manufacturer reserves the right to modify technical data without prior notice. The
Endress+Hauser sales organization will supply you with current information and updates
to these instructions.

2

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Table of contents

Table of contents
5.3.3

5.4

Integrated antenna, PEEK
20 mm (0.75 in) . . . . . . . . . . . . . . . . 18
5.3.4 Integrated antenna, PEEK
40 mm (1.5 in) . . . . . . . . . . . . . . . . . 18
5.3.5 Turning the housing . . . . . . . . . . . . . 19
5.3.6 Turning the display module . . . . . . . . 19
5.3.7 Changing the installation position of
the display module . . . . . . . . . . . . . . 20
5.3.8 Closing the housing covers . . . . . . . . . 22
Post-mounting check . . . . . . . . . . . . . . . . . . . 22

6

Electrical connection . . . . . . . . . . . . . . 23

6.1

6.4

Connecting requirements . . . . . . . . . . . . . . . 23
6.1.1 Cover with securing screw . . . . . . . . . 23
6.1.2 Potential equalization . . . . . . . . . . . . 23
Connecting the device . . . . . . . . . . . . . . . . . . 24
6.2.1 Supply voltage . . . . . . . . . . . . . . . . . 24
6.2.2 Cable specification . . . . . . . . . . . . . . . 25
6.2.3 Overvoltage protection . . . . . . . . . . . 25
6.2.4 Wiring . . . . . . . . . . . . . . . . . . . . . . . 26
6.2.5 Terminal assignment . . . . . . . . . . . . 27
6.2.6 Cable entries . . . . . . . . . . . . . . . . . . . 28
6.2.7 Available device plugs . . . . . . . . . . . . 28
Ensuring the degree of protection . . . . . . . . . 29
6.3.1 Cable entries . . . . . . . . . . . . . . . . . . . 29
Post-connection check . . . . . . . . . . . . . . . . . . 29

7

Operation options . . . . . . . . . . . . . . . . . 30

7.1
7.2

Overview of operation options . . . . . . . . . . . . 30
Operating keys and DIP switches on theelectronic insert . . . . . . . . . . . . . . . . . . . . . . . 30
Structure and function of the operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
7.3.1 User roles and related accessauthorization . . . . . . . . . . . . . . . . . . 30
Access to operating menu via local display . . . 31
7.4.1 Device display (optional) . . . . . . . . . . 31
7.4.2 Operation via Bluetooth® wirelesstechnology (optional) . . . . . . . . . . . . 31
Access to operating menu via Web browser . . 32
7.5.1 Function scope . . . . . . . . . . . . . . . . . 32
7.5.2 Requirements . . . . . . . . . . . . . . . . . . 33
7.5.3 Establishing a connection . . . . . . . . . 34
7.5.4 User interface . . . . . . . . . . . . . . . . . . 35
7.5.5 Disabling the Web server . . . . . . . . . . 36
7.5.6 Logging out . . . . . . . . . . . . . . . . . . . . 37
Access to the operating menu via theoperating tool . . . . . . . . . . . . . . . . . . . . . . . . 37
7.6.1 Connecting the operating tool . . . . . . 38
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
7.7.1 Function range . . . . . . . . . . . . . . . . . 39
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . 39
7.8.1 Function scope . . . . . . . . . . . . . . . . . 39

1

About this document . . . . . . . . . . . . . . . . 5

1.1
1.2

1.5

Purpose of this document . . . . . . . . . . . . . . . . . 5
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5
1.2.1 Safety symbols . . . . . . . . . . . . . . . . . . 5
1.2.2 Symbols for certain types of
information and graphics . . . . . . . . . . . 5
Terms and abbreviations . . . . . . . . . . . . . . . . . 6
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 6
1.4.1 Technical Information (TI) . . . . . . . . . 6
1.4.2 Brief Operating Instructions (KA) . . . . . 6
1.4.3 Safety Instructions (XA) . . . . . . . . . . . . 6
Registered trademarks . . . . . . . . . . . . . . . . . . . 6

2

Basic safety instructions . . . . . . . . . . . . 8

2.1
2.2
2.3
2.4
2.5
2.6
2.7

Requirements for personnel . . . . . . . . . . . . . . . 8
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . 9
Operational safety . . . . . . . . . . . . . . . . . . . . . . 9
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . 9
IT security . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Device-specific IT security . . . . . . . . . . . . . . . 10
2.7.1 Protecting access via a password . . . . 10
2.7.2 Access via Web server . . . . . . . . . . . . 10

6.3

3

Product description . . . . . . . . . . . . . . . . 12

3.1

Product design . . . . . . . . . . . . . . . . . . . . . . . . 12

4

Incoming acceptance and productidentification . . . . . . . . . . . . . . . . . . . . . 13

4.1
4.2

Incoming acceptance . . . . . . . . . . . . . . . . . . . 13
Product identification . . . . . . . . . . . . . . . . . . . 13
4.2.1 Nameplate . . . . . . . . . . . . . . . . . . . . 13
4.2.2 Manufacturer address . . . . . . . . . . . . 14
Storage and transport . . . . . . . . . . . . . . . . . . 14
4.3.1 Storage conditions . . . . . . . . . . . . . . . 14
4.3.2 Transporting the product to themeasuring point . . . . . . . . . . . . . . . . 14

7.3

1.3
1.4

4.3

5

Mounting . . . . . . . . . . . . . . . . . . . . . . . . . 15

5.1
5.2

General instructions . . . . . . . . . . . . . . . . . . . . 15
Mounting requirements . . . . . . . . . . . . . . . . . 15
5.2.1 Internal vessel fittings . . . . . . . . . . . . 15
5.2.2 Avoiding interference echoes . . . . . . . 16
5.2.3 Vertical alignment of antenna axis . . . 16
5.2.4 Radial alignment of the antenna . . . . 16
5.2.5 Optimization options . . . . . . . . . . . . . 16
Mounting the device . . . . . . . . . . . . . . . . . . . 16
5.3.1 Encapsulated antenna, PVDF
40 mm (1.5 in) . . . . . . . . . . . . . . . . . 16
5.3.2 Drip-off antenna PTFE
50 mm (2 in) . . . . . . . . . . . . . . . . . . 17

5.3

Endress+Hauser

6.2

7.4

7.5

7.6
7.7
7.8

3

Table of contents

Micropilot FMR60B PROFINET with Ethernet-APL

8

System integration . . . . . . . . . . . . . . . . 40

11.7

8.1

Overview of device description files . . . . . . . . . 40
8.1.1 Current version data for the device . . . 40
8.1.2 Operating tools . . . . . . . . . . . . . . . . . 40
Device master file (GSD) . . . . . . . . . . . . . . . . . 40
8.2.1 File name of the device master file(GSD) . . . . . . . . . . . . . . . . . . . . . . . . 41
Cyclic data transmission . . . . . . . . . . . . . . . . 42
8.3.1 Overview of the modules . . . . . . . . . . 42
8.3.2 Description of the modules . . . . . . . . 42
8.3.3 Status coding . . . . . . . . . . . . . . . . . . 43
8.3.4 Startup configuration . . . . . . . . . . . . 44
System redundancy S2 . . . . . . . . . . . . . . . . . . 44

12Maintenance . . . . . . . . . . . . . . . . . . . . . . 61

12.1
12.2

Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . 61
Seals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 61

13

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62

8.2
8.3

8.4

9

Commissioning . . . . . . . . . . . . . . . . . . . . 45

9.1
9.2
9.3

9.6
9.7

Preliminaries . . . . . . . . . . . . . . . . . . . . . . . . 45
Function check . . . . . . . . . . . . . . . . . . . . . . . 45
Establishing a connection via FieldCare andDeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . 45
9.3.1 Via PROFINET protocol . . . . . . . . . . . 45
9.3.2 Via service interface (CDI) . . . . . . . . . 46
9.3.3 Hardware settings . . . . . . . . . . . . . . . 46
9.3.4 Configuring the communication
parameters via the software . . . . . . . 46
Configuring the operating language . . . . . . . . 47
9.4.1 Local display . . . . . . . . . . . . . . . . . . . 47
9.4.2 Operating tool . . . . . . . . . . . . . . . . . . 47
9.4.3 Web server . . . . . . . . . . . . . . . . . . . . 47
Configuring the device . . . . . . . . . . . . . . . . . 48
9.5.1 Level measurement in liquids . . . . . . . 48
9.5.2 Commissioning with theCommissioning wizard . . . . . . . . . . . 48
Recording the Echo curve . . . . . . . . . . . . . . . . 49
"Simulation" submenu . . . . . . . . . . . . . . . . . . . 49

10Operation . . . . . . . . . . . . . . . . . . . . . . . . . 50

9.4

9.5

10.1 Reading off the device locking status . . . . . . . 50
10.2 Reading off measured values . . . . . . . . . . . . . 50
10.3 Adapting the device to process conditions . . . . 50

11
11.1

Diagnostics and troubleshooting . . . 51

General troubleshooting . . . . . . . . . . . . . . . . . 51
11.1.1 General errors . . . . . . . . . . . . . . . . . . 51
11.2 Error - SmartBlue operation . . . . . . . . . . . . . . 51
11.3 Diagnostic message . . . . . . . . . . . . . . . . . . . . 52
11.3.1 Diagnostic information on local
display . . . . . . . . . . . . . . . . . . . . . . . 53
11.3.2 Diagnostic information in theoperating tool . . . . . . . . . . . . . . . . . . 53
11.3.3 Status signal . . . . . . . . . . . . . . . . . . . 53
11.3.4 Diagnostic event and event text . . . . . 54
11.4 Remedy informationCalling up . . . . . . . . . . . 54
11.4.1 Graphic display with keys . . . . . . . . . 54
11.4.2 Operating menu . . . . . . . . . . . . . . . . 55
11.5 Adapting the diagnostic information . . . . . . . 55
11.6 List of diagnostic events . . . . . . . . . . . . . . . . . 55

4

Event logbook . . . . . . . . . . . . . . . . . . . . . . . . 57
11.7.1 Event history . . . . . . . . . . . . . . . . . . . 57
11.7.2 Filtering the event logbook . . . . . . . . 58
11.7.3 Overview of information events . . . . . 58
11.8 Resetting the device . . . . . . . . . . . . . . . . . . . . 59
11.8.1 Reset password via operating tool . . . 59
11.8.2 Reset device via operating tool . . . . . 59
11.8.3 Resetting the device via keys on theelectronic insert . . . . . . . . . . . . . . . . 59
11.9 Device information . . . . . . . . . . . . . . . . . . . . 60
11.10 Firmware history . . . . . . . . . . . . . . . . . . . . . . 60

13.1

General information . . . . . . . . . . . . . . . . . . . 62
13.1.1 Repair concept . . . . . . . . . . . . . . . . . 62
13.1.2 Repair of Ex-certified devices . . . . . . . 62
13.2 Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . 62
13.3 Replacement . . . . . . . . . . . . . . . . . . . . . . . . . 62
13.3.1 HistoROM . . . . . . . . . . . . . . . . . . . . . 63
13.4 Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 63
13.5 Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . 63

14

Accessories . . . . . . . . . . . . . . . . . . . . . . . 64

14.1 Weather protection cover 316L . . . . . . . . . . . 64
14.2 Plastic weather protection cover . . . . . . . . . . . 64
14.3 Mounting bracket, adjustable . . . . . . . . . . . . . 65
14.3.1 Dimensions . . . . . . . . . . . . . . . . . . . . 65
14.3.2 Scope of delivery . . . . . . . . . . . . . . . . 66
14.4 M12 socket . . . . . . . . . . . . . . . . . . . . . . . . . . 66
14.5 Remote display FHX50B . . . . . . . . . . . . . . . . . 67
14.6 Gas-tight feedthrough . . . . . . . . . . . . . . . . . . 69
14.7 Field Xpert SMT70 . . . . . . . . . . . . . . . . . . . . . 69
14.8 DeviceCare SFE100 . . . . . . . . . . . . . . . . . . . . 69
14.9 FieldCare SFE500 . . . . . . . . . . . . . . . . . . . . . 69

15

Technical data . . . . . . . . . . . . . . . . . . . . 70

15.1 Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70
15.2 Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
15.3 Environment . . . . . . . . . . . . . . . . . . . . . . . . . 78
15.4 Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 90

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 93

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

1

About this document

1.1

Purpose of this document

About this document

These Operating Instructions contain all the information that is required in the various
phases of the life cycle of the device: from product identification, incoming acceptance andstorage, to mounting, connection, operation and commissioning through totroubleshooting, maintenance and disposal.

1.2

Symbols

1.2.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

1.2.2

Symbols for certain types of information and graphics

Permitted
Procedures, processes or actions that are permitted
Forbidden
Procedures, processes or actions that are forbidden
Tip
Indicates additional information
Reference to documentationA

Reference to page
Reference to graphic
Notice or individual step to be observed
1. , 2. , 3.
Series of steps

Result of a step
1, 2, 3, ...
Item numbers
A, B, C, ...
Views

Endress+Hauser

5About this document

Micropilot FMR60B PROFINET with Ethernet-APL

1.3

Terms and abbreviations

MWP
Maximum working pressure
The MWP is indicated on the nameplate.
εr (Dk value)
Relative dielectric constant
CDI
Common Data Interface

1.4

Documentation

The following document types are available in the Downloads section of the Endress+Hauser website (www.endress.com/downloads):
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• Device Viewer (www.endress.com/deviceviewer): Enter the serial number from thenameplate• Endress+Hauser Operations app: Enter serial number from nameplate or scanmatrix code on nameplate.

1.4.1

Technical Information (TI)

Planning aid
The document contains all the technical data on the device and provides an overview of
the accessories and other products that can be ordered for the device.

1.4.2

Brief Operating Instructions (KA)

Guide that takes you quickly to the 1st measured value
The Brief Operating Instructions contain all the essential information from incoming
acceptance to initial commissioning.

1.4.3

Safety Instructions (XA)

Depending on the approval, the following Safety Instructions (XA) are supplied with thedevice. They are an integral part of the Operating Instructions.
The nameplate indicates the Safety Instructions (XA) that are relevant to the device.

1.5

Registered trademarks

PROFINET®
Registered trademark of the PROFIBUS User Organization, Karlsruhe, Germany
Bluetooth®
The Bluetooth® word mark and logos are registered trademarks owned by the Bluetooth
SIG, Inc. and any use of such marks by Endress+Hauser is under license. Other trademarks
and trade names are those of their respective owners.
Apple®
Apple, the Apple logo, iPhone, and iPod touch are trademarks of Apple Inc., registered in
the U.S. and other countries. App Store is a service mark of Apple Inc.
Android®
Android, Google Play and the Google Play logo are trademarks of Google Inc.

6

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

About this document

KALREZ®, VITON®
Registered trademarks of DuPont Performance Elastomers L.L.C., Wilmington, DE USA

Endress+Hauser

7Basic safety instructions

Micropilot FMR60B PROFINET with Ethernet-APL

2

Basic safety instructions

2.1

Requirements for personnel

The personnel for installation, commissioning, diagnostics and maintenance must fulfill
the following requirements:‣ Trained, qualified specialists must have a relevant qualification for this specific function
and task.
‣ Personnel must be authorized by the plant owner/operator.
‣ Be familiar with federal/national regulations.
‣ Before starting work: personnel must read and understand the instructions in the
manual and supplementary documentation as well as the certificates (depending on
the application).
‣ Personnel must follow instructions and comply with general policies.
The operating personnel must fulfill the following requirements:‣ Personnel are instructed and authorized according to the requirements of the task by
the facility's owner-operator.

‣ Personnel follow the instructions in this manual.

2.2

Intended use

Application and media
The measuring device described in these Operating Instructions is intended for continuous,
non-contact level measurement in liquids, pastes and sludges. Because of its operating
frequency of approx. 80 GHz, a maximum radiated peak power of 6.3 mW and an average
output power of 63 µW, unrestricted use outside of closed, metallic vessels is also
permitted (for example over basins or open channels). Operation does not pose any danger
whatsoever to humans and animals.
If the limit values specified in the "Technical data" and the conditions listed in the
instructions and additional documentation are observed, the measuring device may be
used for the following measurements only:‣ Measured process variables: level, distance, signal strength‣ Calculated process variables: volume or mass in vessels of any shape; flow throughmeasuring weirs or channels (calculated from the level by the linearization
functionality)
To ensure that the measuring device remains in proper condition for the operation time:‣ Use the measuring device only for media against which the process-wetted materials
are adequately resistant.

‣ Observe the limit values in the "Technical data".
Incorrect use
The manufacturer is not liable for damage caused by improper or non-intended use.
Avoid mechanical damage:‣ Do not touch or clean device surfaces with pointed or hard objects.
Clarification of borderline cases:‣ For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistancein verifying the corrosion resistance of fluid-wetted materials, but does not accept any
warranty or liability.

Residual risks
Due to heat transfer from the process as well as power loss in the electronics, thetemperature of the electronics housing and the assemblies it contains (e.g. display module,
main electronics module and I/O electronics module) may rise to 80 °C (176 °F). When in
operation, the sensor may reach a temperature close to the medium temperature.

8

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Basic safety instructions

Danger of burns from contact with surfaces!
‣ For elevated fluid temperatures, ensure protection against contact to prevent burns.

2.3

Workplace safety

When working on and with the device:‣ Wear the required personal protective equipment according to federal/national
regulations.
‣ Switch off the supply voltage before connecting the device.

2.4

Operational safety

Risk of injury!
‣ Operate the device only if it is in proper technical condition, free from errors and faults.
‣ The operator is responsible for the interference-free operation of the device.
Modifications to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers:‣ If modifications are nevertheless required, consult with the manufacturer.
RepairTo ensure continued operational safety and reliability:‣ Carry out repairs on the device only if they are expressly permitted.
‣ Observe federal/national regulations pertaining to the repair of an electrical device.
‣ Use only original spare parts and accessories from the manufacturer.
Hazardous area
To eliminate danger to persons or the installation when the device is used in thehazardous area (e.g. explosion protection, pressure vessel safety):‣ Check the nameplate to verify whether the ordered device can be put to its intended
use in the hazardous area.
‣ Observe the specifications in the separate supplementary documentation, which is an
integral part of this manual.

2.5

Product safety

This device is designed in accordance with good engineering practice to meet state-of-theart safety requirements, has been tested, and left the factory in a condition in which it issafe to operate.
It meets the general safety standards and legal requirements. It also complies with the EU
directives listed in the device-specific EU Declaration of Conformity. Endress+Hauser
confirms this by affixing the CE mark to the device.

2.6

IT security

We only provide a warranty if the device is installed and used as described in the
Operating Instructions. The device is equipped with security mechanisms to protect itagainst any inadvertent changes to the device settings. IT security measures in line with
operators' security standards and designed to provide additional protection for the device
and device data transfer must be implemented by the operators themselves.

Endress+Hauser

9Basic safety instructions

Micropilot FMR60B PROFINET with Ethernet-APL

2.7

Device-specific IT security

The device offers specific functions to support protective measures by the operator. These
functions can be configured by the user and guarantee greater in-operation safety if usedcorrectly. An overview of the most important functions is provided in the following section:• Write protection via hardware write protection switch• Access code to change user role (applies to operation via display, Bluetooth or FieldCare,
DeviceCare, asset management tools (e.g., AMS, PDM and Web server)
Function/interface

Factory setting

Recommendation

Access code(also applies to Web server login or FieldCare connection)

Not enabled (0000)

Assign a customized access code during commissioning

Web server

EnabledOn an individual basis following risk assessment

Service interface (CDI)

EnabledOn an individual basis following risk assessment

Write protection via hardware write protection switch

Not enabled

On an individual basis following risk assessment

2.7.1

Protecting access via a password

Different passwords are available to protect write access to the parameters of the device.
Protect write access to the parameters of the device via the local display, Web browser or
operating tool (e.g. FieldCare, DeviceCare). Access authorization is clearly regulated
through the use of a user-specific access code.
User-specific access code
Write access to the parameters of the device via the local display, Web browser or
operating tool (e.g. FieldCare, DeviceCare) can be protected using the editable, userspecific access code.
When delivered, the device does not have an access code and this code corresponds to0000 (open).
General notes on the use of passwords• During commissioning, change the access code used when the device was delivered• When defining and managing the access code, comply with the general rules for thegeneration of a secure password• The user is responsible for managing the access code and for using the code with due
care
For further information, see →

2.7.2

A

Resetting the device

Access via Web serverThanks to the integrated Web server, the device can be operated and configured using a
Web browser and via PROFINET with Ethernet-APL. In addition to the measured values,
status information on the device is displayed and can be used to monitor device health.
Furthermore the device data can be managed and the network parameters can be
configured.
Access to the network is required for the PROFINET with Ethernet-APL connection.

10

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Basic safety instructions

Supported functions
Data exchange between the operating unit (such as a notebook, for example,) andmeasuring device:• Export parameter settings (PDF file, create documentation of the measuring point
configuration)• Export the Heartbeat Technology verification report (PDF file, only available with the"Heartbeat Verification" application package)• Download driver (GSDML) for system integration
The Web server is enabled when the device is delivered. The Web server can be disabled
via the Web server functionality parameter if necessary (e.g., after commissioning).
The device and status information can be hidden on the login page. This prevents
unauthorized access to the information.
Description of device parameters.

Endress+Hauser

11

Product description

Micropilot FMR60B PROFINET with Ethernet-APL

3

Product description

3.1

Product design

1

2
4
3
5

A0046661

1
1
2
3
4
5

12

Design of the Micropilot FMR60B
Electronics housing
Encapsulated antenna, PVDF, 40 mm (1.5 in)
Drip-off antenna 50 mm (2 in) - threaded or UNI flange process connection
Integrated antenna, PEEK, 20 mm (0.75 in)Integrated antenna, PEEK, 40 mm (1.5 in)Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Incoming acceptance and product identification

4

Incoming acceptance and productidentification

4.1

Incoming acceptance
DELIVERY NOTE

1=2

A0016870

Check the following during incoming acceptance:• Is the order code on the delivery note (1) identical to the order code on the productsticker (2)?
• Are the goods undamaged?
• Do the data on the nameplate correspond to the order specifications and the delivery
note?
• Is the documentation provided?
• If required (see nameplate): are the Safety Instructions (XA) provided?
If one of these conditions is not met, please contact the manufacturer's sales office.

4.2

Product identification

The following options are available for identification of the device:• Nameplate specifications• Extended order code with breakdown of the device features on the delivery note

‣ Device Viewer(www.endress.com/deviceviewer); manually enter the serial number
from the nameplate.
 All the information about the measuring device is displayed.

‣ Endress+Hauser Operations app; manually enter the serial number indicated on the
nameplate or scan the 2D matrix code on the nameplate.
 All the information about the measuring device is displayed.

4.2.1

Nameplate

The information that is required by law and is relevant to the device is shown on the
nameplate, e.g.:• Manufacturer identification• Order number, extended order code, serial number• Technical data, degree of protection• Firmware version, hardware version• Approval-related information, reference to Safety Instructions (XA)
• DataMatrix code (information about the device)

Endress+Hauser

13

Incoming acceptance and product identification

4.2.2

Micropilot FMR60B PROFINET with Ethernet-APL

Manufacturer address

Endress+Hauser SE+Co. KG
Hauptstraße 179689 Maulburg, Germany
Place of manufacture: See nameplate.

4.3

Storage and transport

4.3.1

Storage conditions• Use the original packaging• Store the device in clean and dry conditions and protect from damage caused by shocks
Storage temperature range
See Technical Information.

4.3.2

Transporting the product to the measuring point

LWARNING
Incorrect transport!
The housing or sensor can be damaged or pull off. Risk of injury!
‣ Transport the device to the measuring point in its original packaging or by the processconnection.
‣ Always secure lifting equipment (slings, eyes, etc.) at the process connection and neverlift the device by the electronic housing or sensor. Pay attention to the center of gravityof the device so that it does not tilt or slip unintentionally.

14

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Mounting

5

Mounting

5.1

General instructions

LWARNING
Loss of protection rating if the device is opened in a wet environment.
‣ Only open the device in a dry environment!
1.

Install the device or turn the housing so that the cable entries do not point upwards.
A0029263

2.

Always firmly tighten the housing cover and the cable entries.

3.

Counter-tighten the cable entries.

4.

A drip loop must be provided when laying the cables.

5.2

Mounting requirements

5.2.1

Internal vessel fittingsαA0031777

Avoid internal fittings (point level switches, temperature sensors, struts, vacuum rings,
heating coils, baffles etc.) inside the signal beam. Pay attention to the beam angle α.

Endress+Hauser

15

MountingMicropilot FMR60B PROFINET with Ethernet-APL

5.2.2

Avoiding interference echoes

A0031813

Metal deflector plates, installed at an angle to scatter the radar signals, help prevent
interference echoes.

5.2.3

Vertical alignment of antenna axis

Align the antenna so that it is perpendicular to the product surface.
The maximum reach of the antenna can be reduced, or additional interference signals
can occur, if the antenna is not installed perpendicular to the product.

5.2.4

Radial alignment of the antenna

Based on the directional characteristic, radial alignment of the antenna is not necessary.

5.2.5

Optimization options

Mapping
Measurement can be optimized by electronically suppressing interference echoes.
See the Confirm distance parameter.

5.3

Mounting the device

5.3.1

Encapsulated antenna, PVDF 40 mm (1.5 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 50 mm• Maximum permissible torque: 30 Nm (22 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

16

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Mounting

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D
D

Hmax

50 to 80 mm (2 to 3.2 in)

600 mm (24 in)

80 to 100 mm (3.2 to 4 in)

1 000 mm (24 in)

100 to 150 mm (4 to 6 in)

1 250 mm (50 in)≥ 150 mm (6 in)

1 850 mm (74 in)øDIn the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

5.3.2

Drip-off antenna PTFE 50 mm (2 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 55 mm• Maximum permissible torque: 50 Nm (36 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D
D

Hmax

50 to 80 mm (2 to 3.2 in)

750 mm (30 in)

80 to 100 mm (3.2 to 4 in)

1 150 mm (46 in)

100 to 150 mm (4 to 6 in)

1 450 mm (58 in)≥150 mm (6 in)

2 200 mm (88 in)øDIn the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

Endress+Hauser

17

MountingMicropilot FMR60B PROFINET with Ethernet-APL

5.3.3

Integrated antenna, PEEK 20 mm (0.75 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 36 mm• Maximum permissible torque: 50 Nm (36 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D

øD

D

Hmax

40 to 50 mm (1.6 to 2 in)

200 mm (8 in)

50 to 80 mm (2 to 3.2 in)

300 mm (12 in)

80 to 100 mm (3.2 to 4 in)

450 mm (18 in)

100 to 150 mm (4 to 6 in)

550 mm (22 in)≥ 150 mm (6 in)

850 mm (34 in)In the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

5.3.4

Integrated antenna, PEEK 40 mm (1.5 in)Information concerning threaded connections• When screwing in, turn by the hex bolt only.
• Tool: open-ended wrench 55 mm• Maximum permissible torque: 50 Nm (36 lbf ft)
Information about the mounting nozzle
The maximum nozzle length Hmax depends on the nozzle diameter D.

H max

Maximum nozzle length Hmax as a function of the nozzle diameter D
ØD

Hmax

40 to 50 mm (1.6 to 2 in)

400 mm (16 in)

50 to 80 mm (2 to 3.2 in)

550 mm (22 in)

80 to 100 mm (3.2 to 4 in)

850 mm (34 in)øD

18Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Mounting

ØD

Hmax

100 to 150 mm (4 to 6 in)

1 050 mm (42 in)≥150 mm (6 in)

1 600 mm (64 in)In the case of longer nozzles, reduced measuring performance must be expected.
Please note the following:• The end of the nozzle must be smooth and free from burrs.
• The edge of the nozzle should be rounded.
• Mapping must be performed.
• Please contact the manufacturer's support department for applications with nozzles
that are higher than indicated in the table.

5.3.5

Turning the housing

The housing can be rotated up to 380° by loosening the locking screw.
Your benefits• Easy installation due to optimum alignment of housing• Easily accessible device operation• Optimum readability of the local display (optional)
B

A

C

1

E

D

1

1
A0046660

A
B
C
D
E
1

Plastic single compartment housing (no locking screw)
Aluminum single compartment housing
Single compartment housing, 316L, hygiene (no locking screw)
Dual compartment housing
Dual compartment housing, L-form
Locking screw

NOTICEThe housing cannot be unscrewed fully.
‣ Loosen the external locking screw by a maximum of 1.5 turns. If the screw isunscrewed too much or completely (beyond the screw anchor point), small parts(counter disk) can become loose and fall out.
Tightenthe securing screw (hexagon socket 4 mm (0.16 in)) with maximum‣
3.5 Nm (2.58 lbf ft) ± 0.3 Nm (0.22 lbf ft).

5.3.6

Turning the display module

LWARNING
Supply voltage switched on!
Risk of electric shock and/or explosion!
‣ Switch off the supply voltage before opening the measuring device.

Endress+Hauser

19

MountingMicropilot FMR60B PROFINET with Ethernet-APL

3.

1.
4 0.7 Nm

2.

4.

A0038224

1.

If fitted: release the screw of the cover lock for the electronics compartment coverusing the Allen key.

2.

Unscrew the electronics compartment cover from the transmitter housing and checkthe cover seal.

3.

Press the release mechanism and remove the display module.

4.

Turn the display module to the desired position: maximum 4 × 90 ° in each direction.
Fit the display module on the electronics compartment in the desired position until itclicks into place. Screw the electronics compartment cover back onto the transmitter
housing. If fitted: tighten the screw of the cover lock using the Allen key
0.7 Nm (0.52 lbf ft)±0.2 Nm (0.15 lbf ft).

5.3.7

Changing the installation position of the display module

The installation position of the display can be changed in the case of the dual
compartment housing, L-form.

A0048401

20

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Mounting

LWARNING
Supply voltage switched on!
Risk of electric shock and/or explosion!
‣ Switch off the supply voltage before opening the measuring device.
1→

2→

3→

4 mm

1.

2.
2.

2x

1.
0.7 Nm

A0046831

A0046832

A0046833‣ If fitted: release the screw of‣ Unscrew the display cover and‣ Press the release mechanism,

4→

5→

6→

the cover lock for the display
cover using the Allen key.

check the cover seal.

remove the display module.

4 mm
2x

1.
0.7 Nm2.
A0046834‣ Release the connection.

Endress+Hauser

A0046923‣ If fitted: release the screw of

the cover lock for theconnection compartment coverusing the Allen key.

A0046924‣ Unscrew the connection

compartment cover, check thecover seal. Screw this cover
onto the electronics
compartment instead of thedisplay cover. If fitted: tighten
the screw of the cover lockusing the Allen key

21

MountingMicropilot FMR60B PROFINET with Ethernet-APL

7→

8

Di

sp

lay

2.

1.
A0048406‣ Plug in the connection for the
display module in theconnection compartment.
‣ Fit the display module in thedesired position until it clicks
into place.

5.3.8

A0046928‣ Screw the display cover firmly

back onto the housing. If fitted:
tighten the screw of the coverlock using the Allen key
0.7 Nm (0.52 lbf ft).

Closing the housing covers

NOTICE
Thread and housing damaged from dirt!
‣ Remove dirt (e.g. sand) on the cover and housing thread.
‣ If you encounter resistance when closing the cover, check the thread for dirt and cleanit.
Housing thread
The threads of the electronics and connection compartment can be coated with ananti-friction coating.
The following applies for all housing materials:
Do not lubricate the housing threads.

5.4

Post-mounting checkIs the device free from damage (visual inspection)?



Are the measuring point identification and labeling correct (visual inspection)?

Is the measuring device protected against precipitation and sunlight?



Are the securing screws and cover lock tightened securely?

 Does the measuring device comply with the measuring point specifications?
For example:•  Process temperature•  Process pressure•  Ambient temperature•  Measuring range

22

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

6

Electrical connection

6.1

Connecting requirements

6.1.1

Cover with securing screw

Electrical connection

In the case of devices for use in the hazardous area with a certain type of protection, thecover is sealed by a securing screw.
NOTICE
If the securing screw is not positioned correctly, the cover cannot provide securesealing.
‣ Open the cover: slacken the screw of the cover lock with a maximum of 2 turns so thatthe screw does not fall out. Fit the cover and check the cover seal.
‣ Close the cover: screw the cover securely onto the housing, making sure that the
securing screw is positioned correctly. There should not be any gap between the coverand housing.

2x4
0.7 Nm

A0039520

2

Cover with securing screw

7x4
0.7 Nm

A0050983

3

6.1.2

Cover with securing screw; hygiene housing (only for dust explosion protection)

Potential equalization

The protective ground on the device must not be connected. If necessary, the potential
matching line can be connected to the outer ground terminal of the transmitter before thedevice is connected.

Endress+Hauser

23

Electrical connection

Micropilot FMR60B PROFINET with Ethernet-APL

B

A

1

1

D

C

1

1

E

1
A0046583

A
B
C
D
E
1

Single compartment housing, plastic
Single compartment housing, aluminum
Single compartment housing, 316L hygiene (Ex device)
Dual compartment housing
Dual compartment housing, L-form
Ground terminal for connecting the potential matching line

LWARNING
Explosion Hazard!
‣ Please refer to the separate documentation on applications in hazardous areas for thesafety instructions.
For optimum electromagnetic compatibility:• Keep the potential matching line as short as possible• Observe a cross-section of at least 2.5 mm2 (14 AWG)

6.2

Connecting the device
C

B

A
1

1

D
1

E
1

1

A0046659

A
B
C
D
E
1

Plastic single compartment housing
Aluminum single compartment housing
Single compartment housing, 316L hygiene
Dual compartment housing
Dual compartment housing, L-form
Connection compartment coverDevices with a single compartment housing, 316L hygiene, and conduit gland must beconnected as end-of-line devices. Only one conduit entry must be used.
Housing thread
The threads of the electronics and connection compartment can be coated with ananti-friction coating.
The following applies for all housing materials:
Do not lubricate the housing threads.

6.2.1

Supply voltage

APL performance class A (9.6 to 15 VDC 540 mW)
The APL field switch must be tested to ensure it meets safety requirements (e.g.,
PELV, SELV, Class 2) and must comply with the relevant protocol specifications.

24

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

6.2.2

Electrical connection

Cable specification

Rated cross-section• Supply voltage
0.5 to 2.5 mm2 (20 to 13 AWG)
• Protective earth or grounding of the cable shield> 1 mm2 (17 AWG)
• External ground terminal
0.5 to 4 mm2 (20 to 12 AWG)
Cable outer diameter
The cable outer diameter depends on the cable gland used
• Coupling, plastic:⌀5 to 10 mm (0.2 to 0.38 in)• Coupling, nickel-plated brass:⌀7 to 10.5 mm (0.28 to 0.41 in)• Coupling, stainless steel:⌀7 to 12 mm (0.28 to 0.47 in)
PROFINET with Ethernet-APL
The reference cable type for APL segments is fieldbus cable type A, MAU type 1 and 3(specified in IEC 61158-2). This cable meets the requirements for intrinsically safe
applications according to IEC TS 60079-47 and can also be used in non-intrinsically safe
applications.
Cable typeA

Cable capacitance

45 to 200 nF/km

Loop resistance

15 to 150 Ω/km

Cable inductance

0.4 to 1 mH/km

Further details are provided in the Ethernet-APL Engineering Guideline(https://www.ethernet-apl.org).

6.2.3

Overvoltage protection

The overvoltage protection can optionally be ordered as a "Mounted accessory" via the
product structure
Devices without optional overvoltage protection
The equipment fulfills the requirements of the product standard IEC / DIN EN 61326-1(Table 2 Industrial Environment).
Depending on the type of port (DC power supply, input/output port) different testing levels
according to IEC / DIN EN 61326-1 against transient overvoltages (Surge) are applied(IEC / DIN EN 61000-4-5 Surge):
Test level on DC power ports and input/output ports is 1 000 V line to earth
Devices with optional overvoltage protection• Spark-over voltage: min. 400 VDC• Tested according to IEC / DIN EN 60079-14 sub chapter 12.3 (IEC / DIN EN 60060-1
chapter 7)• Nominal discharge current: 10 kA
NOTICE
Device could be destroyed‣ Always ground device with integrated overvoltage protection.
Endress+Hauser

25

Electrical connection

Micropilot FMR60B PROFINET with Ethernet-APL

Overvoltage category
Overvoltage category II

6.2.4

Wiring

LWARNING
Supply voltage might be connected!
Risk of electric shock and/or explosion!
‣ If the device is used in hazardous areas, make sure to comply with national standards
and the specifications in the Safety Instructions (XAs). The specified cable gland must
be used.
‣ The supply voltage must match the specifications on the nameplate.
‣ Switch off the supply voltage before connecting the device.
‣ If necessary, the potential matching line can be connected to the outer ground terminalof the device before the power supply lines are connected.
‣ A suitable circuit breaker should be provided for the device in accordance with IEC/EN
61010.
‣ The cables must be adequately insulated, with due consideration given to the supply
voltage and the overvoltage category.
‣ The connecting cables must offer adequate temperature stability, with due
consideration given to the ambient temperature.
‣ Only operate the measuring device with the covers closed.
Connect the device in the following order:

26

1.

Release the cover lock (if provided).

2.

Unscrew the cover.

3.

Guide the cables into the cable glands or cable entries.

4.

Connect the cables.

5.

Tighten the cable glands or cable entries so that they are leak-tight. Counter-tighten
the housing entry.

6.

Screw the cover securely back onto the connection compartment.

7.

If provided: tighten the screw of the cover lock using the Allen key
0.7 Nm (0.52 lbf ft) ±0.2 Nm (0.15 lbf ft).

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

6.2.5

Electrical connection

Terminal assignment

Single compartment housing

3
2

-

1

+

A00425944123Connection terminals and ground terminal in the connection compartment
Positive terminal
Negative terminalInternal ground terminal

Dual compartment housing

- +

3

2 1
A0042803

51
2
3Endress+Hauser

Connection terminals and ground terminal in the connection compartment
Positive terminal
Negative terminalInternal ground terminal

27

Electrical connection

Micropilot FMR60B PROFINET with Ethernet-APL

Dual compartment housing, L-form

- +

3

2 1
A00458426123Connection terminals and ground terminal in the connection compartment
Positive terminal
Negative terminalInternal ground terminal

6.2.6

Cable entries
C

B

A
1

D

E

1

1

1

1

2

2

2

2

2
A0046584

A
B
C
D
E
1
2

Single compartment housing, plastic
Single compartment housing, aluminum
Single compartment housing, 316L hygiene
Dual compartment housing
Dual compartment housing, L-form
Cable entry
Dummy plug

The type of cable entry depends on the device version ordered.
Always route connecting cables downwards so that moisture cannot penetrate theconnection compartment.
If necessary, create a drip loop or use a weather protection cover.

6.2.7

Available device plugs

In the case of devices with a plug, it is not necessary to open the housing forconnection purposes.
Use the enclosed seals to prevent the penetration of moisture into the device.

28

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Electrical connection

Devices with M12 plug

4

3

1

2
A0011175

7
1
2
3
4

View of the plug-in connection on the device
APL signal APL signal +Shielding
Not assigned

Various M12 sockets are available as accessories for devices with M12 plugs.

6.3

Ensuring the degree of protection

6.3.1

Cable entries• M20 coupling, plastic, IP66/68 NEMA TYPE 4X/6P• M20 coupling, nickel-plated brass, IP66/68 NEMA TYPE 4X/6P• M20 coupling, 316L, IP66/68 NEMA TYPE 4X/6P• M20 thread, IP66/68 NEMA Type 4X/6P• G1/2 thread, IP66/68 NEMA Type 4X/6P
If the G1/2 thread is selected, the device is delivered with an M20 thread as standard
and a G1/2 adapter is included with the delivery, along with the corresponding
documentation• NPT 1/2 thread, IP66/68 NEMA Type 4X/6P• Dummy plug transport protection: IP22, NEMA TYPE 2• M12 plug
• When housing is closed and connecting cable is plugged in: IP66/67, NEMA Type 4X• When housing is open or connecting cable is not plugged in: IP20, NEMA Type 1
NOTICEM12 plug: Loss of IP protection class due to incorrect installation!
‣ The degree of protection only applies if the connecting cable used is plugged in andscrewed tight.
The
degree of protection only applies if the connecting cable used is specified according‣
to IP67, NEMA Type 4X.
‣ The IP protection classes are only maintained if the dummy cap is used or the cable isconnected.

6.4Endress+Hauser

Post-connection check

Are the device or cables undamaged (visual inspection)?



Do the cables used comply with the requirements?

Do the mounted cables have strain relief?



Cable glands mounted, securely tightened and leak-tight?

Does the supply voltage correspond to the specifications on the nameplate?



No reverse polarity, terminal assignment correct?



Cover screwed down correctly?



Cover lock tightened correctly?
29

Operation options

Micropilot FMR60B PROFINET with Ethernet-APL

7

Operation options

7.1

Overview of operation options• Operation via operating keys and DIP switches on the electronic insert• Operation via optical operating keys on the device display (optional)• Operation via Bluetooth® wireless technology (with optional device display with
Bluetooth) with SmartBlue app or FieldXpert, DeviceCare• Operation via Web server• Operation via operating tool (Endress+Hauser FieldCare/DeviceCare) or FDI Hosts (e.g.,
PDM)

7.2

Operating keys and DIP switches on the electronic
insertNet

Mod

Displayof f on1

3 2
A00460618
1
2
3Operating keys and DIP switches on the Ethernet-APL electronic insert
Operating key for Reset password and Reset device
DIP switch for setting the service IP address
DIP switch for locking and unlocking the device

The setting of the DIP switches on the electronic insert has priority over the settings
made via other operation methods (e.g. FieldCare/DeviceCare).

7.3

Structure and function of the operating menu

The differences between the structure of the operating menus of the local display and theEndress+Hauser FieldCare or DeviceCare operating tools can be summarized as follows:
The local display is suitable for configuring simple applications.
The operating tools (FieldCare, DeviceCare, SmartBlue, AMS, PDM, etc.) can be used toconfigure the parameters of wide-ranging applications.
More elaborate applications can be configured with the Web server.
Wizards help the user to commission the various applications. The user is guided throughthe individual configuration steps.

7.3.1

User roles and related access authorization

The two user roles Operator and Maintenance (as-delivered state) have different write
access to the parameters if a device-specific access code has been defined. This access code
protects the device configuration from unauthorized access.
If an incorrect access code is entered, the user obtains the access rights of the Operator
role.

30

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Operation options

7.4

Access to operating menu via local display

7.4.1

Device display (optional)

Possible to operate the optical operating keys through the cover. No need to open thedevice.
Functions:• Display of measured values and fault and notice messages• Background lighting, which switches from green to red in the event of an error• The device display can be removed for easier operation
Backlighting is switched on or off depending on the supply voltage and the current
consumption.
The device display is also optionally available with Bluetooth® wireless technology.

ESC

-

+

E

1
A0039284

9Graphic display with optical operating keys (1)• Key • Navigate downwards in the picklist• Edit the numerical values or characters within a function• Key • Navigate upwards in the picklist• Edit the numerical values or characters within a function• Key • Change from main display to main menu
• Confirm entry• Jump to the next item
• Select a menu item and activate the edit mode
• Unlock/lock the display operation• Press and hold the  key to display a short description of the selected parameter (ifavailable)•  key and  key (ESC function)• Exit edit mode for a parameter without saving the changed value• Menu at a selection level: pressing the keys simultaneously takes the user back up alevel in the menu
• Press and hold the keys simultaneously to return to the upper level

7.4.2

Operation via Bluetooth® wireless technology (optional)

Prerequisite• Device with device display including Bluetooth• Smartphone or tablet with Endress+Hauser SmartBlue app or PC with DeviceCare fromversion 1.07.05 or FieldXpert SMT70

Endress+Hauser

31Operation options

Micropilot FMR60B PROFINET with Ethernet-APL

The connection has a range of up to 25 m (82 ft). The range can vary depending on
environmental conditions such as attachments, walls or ceilings.
The operating keys on the display are locked as soon as the device is connected via
Bluetooth.
A flashing Bluetooth symbol indicates that a Bluetooth connection is available.
SmartBlue app
1.

Scan the QR code or enter "SmartBlue" in the search field of the App Store or GooglePlay.
A0039186

2.

Start the SmartBlue app.

3.

Select device from livelist displayed.

4.

Login:
 Enter the user name: admin
Password: serial number of the device.

5.

Change the password after logging in for the first time!

Prerequisites
System requirements
The SmartBlue app is available to download for smartphones or tablets.
Please see the "App Store (Apple)" or "Google Play Store" for information regarding thecompatibility of the SmartBlue app with mobile terminals.
Initial password
The serial number of the device serves as the initial password when the connection isestablished for the first time.
Please note the following
If the Bluetooth display is removed from one device and installed in another device:• All the log-in data are only saved in the Bluetooth display and not in the device• The password changed by the user is also saved in the Bluetooth display

7.5

Access to operating menu via Web browser

7.5.1

Function scopeThanks to the integrated Web server the device can be operated and configured via a Web
browser. The structure of the operating menu is the same as for the local display. In
addition to the measured values, device status information is also displayed and allows
users to monitor the status of the device. Furthermore the device data can be managed
and the network parameters can be configured.

32

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

7.5.2

Operation options

Requirements

Computer software
Recommended operating systems• Microsoft Windows 7 or higher.
• Mobile operating systems:• iOS• Android
Microsoft Windows XP is supported.
Web browsers supported• Microsoft Internet Explorer 8 or higher• Microsoft Edge
• Mozilla Firefox• Google Chrome• SafariComputer settings
User rights
Corresponding user rights (e.g., administrator rights) for TCP/IP and proxy server settings
are required (for changing the IP address, subnet mask etc.).
Proxy server settings of the Web browser
The Web browser Use proxy server for LAN setting must be disabled.
JavaScript
JavaScript must be enabled.
When installing a new firmware version: To enable correct data display, clear the
temporary memory (cache) of the Web browser under Internet options.

Endress+Hauser

33

Operation options

Micropilot FMR60B PROFINET with Ethernet-APL

7.5.3

Establishing a connection

Via PROFINET with Ethernet-APL network
1

2

3

4

5

6

6

6

6
A0046097

 10
1
2
3
4
5
6

Options for remote operation via PROFINET with Ethernet-APL network: star topology

Automation system, e.g., Simatic S7 (Siemens)
Ethernet switch
Computer with Web browser (e.g., Microsoft Edge) for accessing the integrated device Web server orcomputer with operating tool (e.g., FieldCare, DeviceCare, SIMATIC PDM) with iDTM Profinet Communication
APL power switch (optional)
APL field switch
APL field device

Call up the website via the computer in the network. The IP address of the device must be
known.
The IP address can be assigned to the device in a variety of ways:• Dynamic Configuration Protocol (DCP), factory setting
The automation system (e.g., Siemens S7) automatically assigns the IP address to thedevice• Software addressing
The IP address is entered via the IP address parameter• DIP switch for service
The device then has the fixed IP address 192.168.1.212
The IP address is only adopted following a restart.
The IP address can now be used to establish the connection to the network
The default setting is that the device uses the Dynamic Configuration Protocol (DCP). The
automation system (e.g., Siemens S7) automatically assigns the IP of the device.
Starting the Web browser and logging in

34

1.

Start the Web browser on the computer.

2.

Enter the IP address of the device in the address line of the Web browser
 The login page appears.

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

1

2 3

Operation options

4

5

6

7
8
A0046626 11

Web browser login

1
2
3
4
5
6
7
8

Device tag
Device name
Status signal
Locking status
Current measured values
Select the language
Enter the "Password" parameter
Login

1.

Select the preferred Language parameter for the Web browser.

2.

Enter the Password parameter (factory setting 0000).

3.

Confirm entry with Login .

7.5.4

User interface

1

3

2

A0052432

 12
1
2
3

Endress+Hauser

User interface with sample contents

HeaderWorking area
Navigation area

35

Operation options

Micropilot FMR60B PROFINET with Ethernet-APL

Header
The following information appears in the header:• Device tag• Device name• Status signal• Locking status• Current measured values
Navigation area
If a function is selected in the function bar, the submenus of the function open in thenavigation area. The user can now navigate within the menu structure.
Working area
Depending on the selected function and the related submenus, various actions can be
performed in this area:• Configure parameters• Read measured values• Call up help text
Adopting a value

 13
1

Example of Enter buttonEnter button in the operating tool

The value entered is only adopted by pressing the Enter key or clicking on the Enter button
(1).

7.5.5

Disabling the Web server

The Web server of the measuring device can be switched on and off as required using the
Web server functionality parameter.
Navigation"System" menu → Connectivity → Interfaces
Parameter overview with brief description
Parameter
Web server functionality

36

Description
Switch web server on and off, switch off HTML.

Selection• Disable• EnableEndress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Operation options

Range of functions of "Web server functionality" parameter
OptionDescription

Disable• The Web server is completely disabled.
• Port 80 is locked.

Enable• The complete functionality of the Web server is available.
• JavaScript is used.
• The password is transferred in an encrypted state.
• Any change to the password is also transferred in an encrypted state.

Enabling the Web server
If the Web server is disabled, it can only be re-enabled with the Web server functionality
parameter via the following operating options:• Via local display• Via the "FieldCare" operating tool
• Via the "DeviceCare" operating tool
• Via the PROFINET startup record

7.5.6

Logging out

1.

Select the Logout entry in the function bar.
 The home page with the Login box appears.

2.

Close the Web browser.
Once communication with the Web server is established via the standard IP address
192.168.1.212, the DIP switch must be reset (from ON → OFF). Following a reboot,the configured IP address of the device is once again active for network
communication.

7.6

Access to the operating menu via the operating tool

The structure of the operating menu in the operating tools is the same as for operation via
the local display. The range of functions is different however.

Endress+Hauser

37

Operation options

Micropilot FMR60B PROFINET with Ethernet-APL

7.6.1

Connecting the operating tool

Via PROFINET with Ethernet-APL network
1

2

3

4

5

6

6

6

6
A0046097

 14
1
2
3
4
5
6

Options for remote operation via PROFINET with Ethernet-APL network: star topology

Automation system, e.g., Simatic S7 (Siemens)
Ethernet switch
Computer with Web browser (e.g., Microsoft Edge) for accessing the integrated device Web server orcomputer with operating tool (e.g., FieldCare, DeviceCare, SIMATIC PDM) with iDTM Profinet Communication
APL power switch (optional)
APL field switch
APL field device

Call up the website via the computer in the network. The IP address of the device must be
known.
The IP address can be assigned to the device in a variety of ways:• Dynamic Configuration Protocol (DCP), factory setting
The automation system (e.g., Siemens S7) automatically assigns the IP address to thedevice• Software addressing
The IP address is entered via the IP address parameter• DIP switch for service
The device then has the fixed IP address 192.168.1.212
The IP address is only adopted following a restart.
The IP address can now be used to establish the connection to the network
The default setting is that the device uses the Dynamic Configuration Protocol (DCP). The
automation system (e.g., Siemens S7) automatically assigns the IP of the device.

38

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Operation options

Service interface
1

2

3

A00391481
2
3

Computer with FieldCare/DeviceCare operating toolCommubox FXA291
Service interface (CDI) of the device (= Endress+Hauser Common Data Interface)

To update (flash) the device firmware, the device must be powered via the power
supply terminals.

7.7

FieldCare

7.7.1

Function range

FDT-based plant asset management tool from Endress+Hauser. FieldCare can configure all
smart field devices in a system and helps you manage them. By using the status
information, FieldCare is also a simple but effective way of checking their status andcondition.
Access is via:
• CDI service interface• PROFINET interface
Typical functions:• Configuration of transmitter parameters• Loading and saving of device data (upload/download)• Documentation of the measuring point• Visualization of the measured value memory (line recorder) and event logbook
For additional information about FieldCare, see Operating Instructions BA00027S
and BA00059S

7.8

DeviceCare

7.8.1

Function scope

Tool for connecting and configuring Endress+Hauser field devices.
The fastest way to configure Endress+Hauser field devices is with the dedicated"DeviceCare" tool. Together with the device type managers (DTMs), DeviceCare presents a
convenient, comprehensive solution.
For details, see Innovation Brochure IN01047S

Endress+Hauser

39

System integration

Micropilot FMR60B PROFINET with Ethernet-APL

8

System integration

8.1

Overview of device description files

8.1.1

Current version data for the device

Firmware version

01.00.zz• On the title page of the Operating Instructions• On the transmitter nameplate• System → Information → Firmware version

Firmware version parameter release date

04.2023–

Manufacturer ID

ID: 0x11

Guidance → Commissioning → Device identification → Manufacturer ID

Device ID

ID: A1C1

Guidance → Commissioning → Device identification → Device ID
On the transmitter nameplate

Profile 4 device ID

B321

On the transmitter nameplate

Device revision

1

On the transmitter nameplate

PROFINET version

2.4x

–

PA profile version

4.0x

Application → PROFINET → Information → PA profile version

8.1.2

Operating tools

The suitable device description file for the individual operating tools is listed in the tablebelow, along with information on where the file can be acquired.
Operating tool via
Service interface (CDI)

Sources for obtaining device descriptions

FieldCare• www.endress.com → Downloads section• CD–ROM (contact Endress+Hauser)• DVD (contact Endress+Hauser)

DeviceCare• www.endress.com → Downloads section• CD–ROM (contact Endress+Hauser)• DVD (contact Endress+Hauser)

SMT70

Use update function of handheld terminal

AMS Device Manager(Emerson Process Management)www.endress.com → Downloads section

SIMATIC PDM(Siemens)

www.endress.com → Downloads section

8.2

Device master file (GSD)

In order to integrate field devices into a bus system, the PROFINET with Ethernet-APL
system needs a description of the device parameters, such as output data, input data, data
format and data volume.
These data are available in the device master file (GSD) which is provided to theautomation system when the communication system is commissioned. In addition device
bit maps, which appear as icons in the network structure, can also be integrated.
The device master file (GSD) is in XML format, and the file is created in the GSDML
description markup language.
Downloading the device master file (GSD)• Via Web server: Menu path System → Device drivers• Via www.endress.com/download

40

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

8.2.1

System integration

File name of the device master file (GSD)

Example of the name of a device master file:
GSDML-V2.43-EH-Micropilot-202304dd.xml (dd corresponds to the release day)Endress+Hauser

GSDML

Description language

V2.43

Version of the PROFINET specification

EH

Endress+Hauser

Micropilot

Instrument family

20230422

Date of issue (year, month, day)

.xml

File name extension (XML file)41

System integration

Micropilot FMR60B PROFINET with Ethernet-APL

8.3

Cyclic data transmission

8.3.1

Overview of the modules

The following graphic shows which modules are available to the device for cyclic dataexchange. Cyclic data exchange is performed with an automation system.
Application → PROFINET → Analog input → Analog input 1 to 11 → Assign process
variable
DeviceSlot

Direction
Data flow

Analog input Level

1→

Analog input Distance2→

Analog input Volume3→

Analog input Sensor temperature

20→

Analog input Electronics temperature

21→

Analog input Absolute echo amplitude

22→

Analog input Relative echo amplitude

23→

Analog input Percent of range

24→

Analog input (Area of incoupling)25→

Analog input Build-up index

26→

Analog input Foam index

27→

Binary input Heartbeat Technology

80→

Binary input (Build up/Foam)

81→

Binary output Heartbeat Technology

210←

Module8.3.2

Control system

PROFINET

Description of the modules

The data structure is described from the perspective of the automation system:• Input data: are sent from the device to the automation system• Output data: are sent from the automation system to the device
Module: Analog input
Transmission of input variables from the device to the automation system:
Analog input modules cyclically transmit the selected input variables, including the status,
from the device to the automation system. The input variable is depicted in the first four
bytes in the form of a floating point number as per the IEEE 754 standard. The fifth byte
contains status information pertaining to the input variable. The Analog input module
Level in slot 1 is contained in the Level PA PROFILE GSD, the other Analog input modules
can only be used with the manufacturer's GSD.
Module: Binary output
The Binary output module can cyclically receive discrete output values from theautomation system. The device implements an 8-bit type as described in PA PROFILE 4.0x.
Of this, 1 bit is used to signal to the device that Heartbeat Verification should be started.
Bit

Function

Description

0

Start verification

Start the verification

1-7

-

-

42

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

System integration

Module: Binary input
The Binary input module can cyclically send discrete values from the device to theautomation system. The status of the Heartbeat Verification is transmitted for the device:
Module: Binary input Heartbeat Technology slot 80
Bit Function

Description

0

Status parameter Not done option

Verification not carried out

1

Status parameter Failed option

The device has not passed the verification. At least one test group
was outside the specifications.

2

Status parameter Busy option

Verification in progress

3

Status parameterDone option

Verification carried out

4

Verification result parameter Failed option

The device has not passed the verification. At least one test groupis outside the specifications.

5

Verification result parameter Passed option

The device has passed the verification. All of the verified test
groups corresponded to the specifications.

6

The verification result is also "Passed" if the result for an individual test groupis "Failed" and the result for all other test groups is "Passed".

7

Verification result parameter Not done option

Verification not carried out

Module: Binary input Sensor diagnostics slot 81
Bit

Function

Description

0

Buildup on the sensor

Buildup detected on the sensor

1

Foam generation

Foam generation detected

2-7

-

8.3.3
StatusStatus coding
Coding (hex)Meaning

BAD - Maintenance alarm

0x24

No measured value available, as a device error has occurred.

BAD - Process related

0x28

No measured value available, as the process conditions are not within thedevice's technical specification limits.

BAD - Function check

0x3C

A function check is active (e.g. cleaning or calibration)

UNCERTAIN - Initial value

0x4F

A predefined value is output until a correct measured value is available
again or corrective measures have been performed that change thisstatus.

UNCERTAIN - Maintenancedemanded

0x68

Wear and tear has been detected. Maintenance is needed shortly toensure the device remains operational.
The measured value might be invalid. The use of the measured value
depends on the application.

UNCERTAIN - Process related

0x78

The process conditions are not within the device's technical specification
limits. This could have a negative impact on the quality and accuracy ofthe measured value.
The use of the measured value depends on the application.

GOOD - OK

0x80

No error has been diagnosed.

GOOD - Maintenance demanded

0xA8

The measured value is valid.
It is highly advisable to service the device in the near future.

GOOD - Function check

0xBC

The measured value is valid.
The device performs an internal function check. The function check does
not have any noticeable effect on the process.

Endress+Hauser

43System integration

Micropilot FMR60B PROFINET with Ethernet-APL

8.3.4
Startup configuration(NSU)

Startup configuration

The automation system adopts the configuration of the most important parameters of the device.
• Interfaces:• Display operation• Web server functionality• Bluetooth activation• Service (UART-CDI)• System units:• Level unit
• Volume unit
• Temperature unit
• Application:• Maximum value• Intermediate height• Medium type
• Full calibration• Empty calibration• Diameter• Diagnostic settings:
Diagnostic behavior for diverse diagnostic information• Analog inputs:
Damping

8.4

System redundancy S2

A redundant layout with two automation systems is necessary for processes that are incontinuous operation. If one system fails the second system guarantees continued,
uninterrupted operation. The device supports S2 system redundancy and can communicate
simultaneously with both automation systems.
1

3

2

4

5

5
A0046154

 15
1
2
3
4
5

Example of the layout of a redundant system (S2): star topology

Automation system 1
Synchronization of automation systems
Automation system 2
Ethernet-APL Field Switch
DeviceAll the devices in the network must support S2 system redundancy.

44

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

9Commissioning

Commissioning
All configuration tools provide a commissioning assistant that supports the user whensetting the most important configuration parameters (Guidance menu
Commissioning wizard).

9.1

Preliminaries

The measuring range and the unit in which the measured value is transmitted correspondto the data on the nameplate.

9.2

Function check

Before commissioning the measuring point, check whether the post-installation and postconnection checks have been performed.
•→
•→

A
A

Post-installation check
Post-connection check

9.3

Establishing a connection via FieldCare andDeviceCare

9.3.1

Via PROFINET protocol

1

2

3

4

5
A0046623

 16
1
2
3
4
5

Endress+Hauser

Options for remote operation via PROFINET protocol

Computer with Web browser or with operating tool (e.g.,, DeviceCare)
Automation system
Field Xpert SMT70
Mobile handheld terminal
Local operation via display module

45

Commissioning

Micropilot FMR60B PROFINET with Ethernet-APL

9.3.2

Via service interface (CDI)1

2

3A00391481
2
3

Computer with FieldCare/DeviceCare operating toolCommubox FXA291
Service interface (CDI) of the measuring device (= Endress+Hauser Common Data Interface)

9.3.3

Hardware settings

Setting the device nameA measuring point can be quickly identified within a plant on the basis of the Device tag
parameter and PROFINET device name parameter. The Device tag parameter, which is
specified ex works or defined when ordering, can be changed in the operating menu.
Configuring the "Device tag" parameter via the operating menu
The Device tag parameter can be adapted via the operating menu or automation system.
System → Device management
Configuring the "PROFINET device name" parameter via the operating menu
Application → PROFINET → Configuration
Configuring the "PROFINET device name" parameter via the automation system
The PROFINET device name parameter can be adapted individually via the automation
system.
When assigning the PROFINET device name parameter via the automation system:
assign the device name in lower case letters.
Activating the default IP address
Activating the default IP address via the DIP switch
The device can be set to the default IP address 192.168.1.212 via DIP switches.
1.

Set DIP switch 2 on the electronic insert from OFF → ON.

2.

Reconnect the device to the power supply.
 The default IP address is used once the device is restarted.

9.3.4

Configuring the communication parameters via the software• IP address• Subnet mask
• Default gateway
Menu path: System → Connectivity → Ethernet

46

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

9.4

Configuring the operating language

9.4.1

Local display

Commissioning

Setting the operating language
To set the operating language, the display first needs to be unlocked:
1.

Press the  key for at least 2 s.
 A dialog box appears.

2.

Unlock the display operation.

3.

Select the Language parameter in the main menu.

4.

Press the  key.

5.

Select the desired language with the  or  key.

6.

Press the  key.
Display operation locks automatically (except in the Safety mode wizard):• after 1 min on the main page if no key has been pressed• after 10 min within the operating menu if no key has been pressed

9.4.2

Operating toolSet display language
System → Display → Language
Selection in Language parameter; Visibility depends on order options or device settings

9.4.3

Web server

1

A0048882

1

Endress+Hauser

Language setting

47

Commissioning

Micropilot FMR60B PROFINET with Ethernet-APL

9.5

Configuring the device

9.5.1

Level measurement in liquids

A

R

C

L

F

E

D

100%

0%

A0016933

 17
R
A
C
D
L
E
F

Configuration parameters for level measurement in liquids

Reference point of measurement
Length of antenna + 10 mm (0.4 in)
50 to 80 mm (1.97 to 3.15 in); medium εr < 2
Distance
Level"Empty calibration" parameter (= 0 %)"Full calibration" parameter (= 100 %)In the case of media with a low dielectric constant, εr < 2 , the tank floor may be visible
through the medium at very low levels (lower than level C). Reduced accuracy must be
expected in this range. If this is not acceptable, the zero point should be positioned at a
distance C above the tank floor in these applications → Configuration parameters forlevel measurement in liquids.

9.5.2

Commissioning with the Commissioning wizard

In FieldCare, DeviceCare, SmartBlue and on the display, the Commissioning wizard isavailable to guide the user through the initial commissioning steps.
Complete this wizard to commission the device.
For each parameter, enter the appropriate value or select the appropriate option.
NOTE
If you exit the wizard before completing all required parameters, the changes you havemade will be saved. For this reason, the device may then be in an undefined state!
In this case, a reset to the default settings is recommended.

48

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

9.6

Commissioning

Recording the Echo curve

Record the current Echo curve as a reference echo curve for subsequent diagnostic
purposes
After the measurement has been configured, it is recommended to record the current Echo
curve as a reference echo curve.
The Save reference curve parameter in the Echo curve submenu is used to record the
Echo curve.
Diagnostics → Echo curve → Save reference curve‣ Under the Save reference curve parameter, activate the Customer reference curve
option9.7"Simulation" submenu

Simulation of a process variable, a pulse output or a diagnostic event.

Endress+Hauser

49

Operation

Micropilot FMR60B PROFINET with Ethernet-APL

10Operation

10.1

Reading off the device locking status

Displaying active write protection:• In the Locking status parameter
Menu path of local display: at the top operating level
Menu path of operating tool: System → Device management• In the operating tool in the header

10.2

Reading off measured values

Many measured values can be read off in the header of the Web server.
All the measured values can be read off using the Measured values submenu.
Navigation"Application" menu → Measured values

10.3

Adapting the device to process conditions

The following menus are available for this purpose:• Basic settings in the Guidance menu
• Advanced settings in:• Diagnostics menu
• Application menu
• System menu

50

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Diagnostics and troubleshooting

11

Diagnostics and troubleshooting

11.1

General troubleshooting

11.1.1

General errors

Device is not responding• Possible cause: Supply voltage does not match the specification on the nameplate
Remedial action: Apply the correct voltage• Possible cause: The connecting cables are not in contact with the terminals
Remedial action: Check the electrical contact between cables and correct if necessary
Values not visible on the display• Possible cause: Display setting is too bright or too dark
Remedy:Use the Contrast display parameter to increase or reduce contrast
Navigation path: System → Display → Contrast display• Possible cause: The plug of the display cable is not connected correctly
Remedial action: Connect the plug correctly• Possible cause: Display is defective
Remedial action: Replace the display"Communication error" is indicated on the display when the device is started or the
display is connected• Possible cause: Electromagnetic interference influence
Remedial action: Check grounding of the device• Possible cause: Defective cable connection or display plug
Remedial action: Replace the display
Communication via CDI interface not working
Possible cause: Wrong setting of the COM port on the computer
Remedial action: Check the setting of the COM port on the computer and correct it ifnecessary
Device measuring incorrectly
Possible cause: Parameter configuration error
Remedial action: Check and correct the parameter configuration

11.2

Error - SmartBlue operation

Operation via SmartBlue is only possible on devices that have a display with Bluetooth(optionally available).
Device is not visible in the live list
• Possible cause: No Bluetooth connection available
Remedial action: Enable Bluetooth in the field device via display or software tool and/orin the smartphone/tablet• Possible cause: Bluetooth signal outside range
Remedial action: Reduce distance between field device and smartphone/tablet
The connection has a range of up to 25 m (82 ft).
Operating radius with intervisibility 10 m (33 ft)• Possible cause: Geopositioning is not enabled on Android devices or is not permitted for
the SmartBlue app
Remedial action: Enable/permit the geopositioning service on Android device for the
SmartBlue app• Display does not have Bluetooth

Endress+Hauser

51

Diagnostics and troubleshooting

Micropilot FMR60B PROFINET with Ethernet-APL

Device appears in the live list but a connection cannot be established• Possible cause: The device is already connected with another smartphone/tablet via
Bluetooth
Only one point-to-point connection is permitted
Remedial action: Disconnect the smartphone/tablet from the device• Possible cause: Incorrect user name and password
Remedial action: The standard user name is "admin" and the password is the device serial
number indicated on the device nameplate (only if the password was not changed by theuser beforehand)
If the password has been forgotten, contact Endress+Hauser Service(www.addresses.endress.com)
Connection via SmartBlue not possible• Possible cause: Incorrect password entered
Remedial action: Enter the correct password, paying attention to lower/upper case
• Possible cause: Forgotten password
Remedial action: Contact Endress+Hauser Service (www.addresses.endress.com)
Login via SmartBlue not possible• Possible cause: Device is being put into operation for the first time
Remedial action: Enter the user name "admin" and the password (device serial number),paying attention to lower/upper case
• Possible cause: The electric current and voltage are not correct.
Remedial action: Increase the supply voltage.
Device cannot be operated via SmartBlue• Possible cause: Incorrect password entered
Remedial action: Enter the correct password, paying attention to lower/upper case
• Possible cause: Forgotten password
Remedial action: Contact Endress+Hauser Service (www.addresses.endress.com)• Possible cause: Operator option has no authorization
Remedial action: Change to Maintenance option

11.3

Diagnostic message

Faults detected by the self-monitoring system of the device are displayed as a diagnostic
message in alternation with the measured value.
XXXXXXXXX

S

20.50
X

XX

S

XXXXXXXXX

S801
Supply voltageiMenu
A0051136

 18

Diagnostic message displayed in alternating sequence with the measured value

If two or more diagnostic events occur simultaneously, only the diagnostic message withthe highest priority is shown.

52

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

11.3.1

Diagnostics and troubleshooting

Diagnostic information on local display

1

X

2
A0043759 191
2

Segment display without keys

Status symbol for event level
Status signal with diagnostic event

XXXXXXXXXSS801
XXXXXXXXX

1
2
3i
A0043103

 20
1
2
3

Graphic display with keys

Status signal
Status symbol with diagnostic event and preceding symbol for event level
Event text

11.3.2

Diagnostic information in the operating tool

If a diagnostic event has occurred in the device, the status signal appears in the top left
status area of the operating tool together with the corresponding symbol for the event
level according to NAMUR NE 107.
Click the status signal to see the detailed status signal.
Pending diagnostic messages can also be displayed in the Active diagnostics parameter.
The diagnostic events and remedial measures can be printed out in the Diagnostic list
submenu.

11.3.3

Status signal

F
Failure (F)A device error has occurred. The measured value is no longer valid.
C
Function check (C)The device is in the service mode (e.g. during a simulation).
S
Out of specification (S)Device operation:• Outside of the technical specifications (e.g. during startup or a cleaning)• Outside of the configuration performed by the user (e.g. sensor frequency outside theconfigured span)
M
Maintenance required (M)Maintenance required. The measured value is still valid.
Endress+Hauser

53

Diagnostics and troubleshooting

Micropilot FMR60B PROFINET with Ethernet-APL

11.3.4

Diagnostic event and event text

The fault can be identified by means of the diagnostic event. The event text helps you byproviding information about the fault. In addition, the associated status symbol isdisplayed in front of the diagnostic event.
2
1

Diagnostic list
Diagnostics 1

S

3

4
5
6

S801 Supply voltage

Diagnostics 2
Diagnostics 3

7
Supply voltage(ID:203)
S801 0d00h02m25s
Increase supply voltage

8A0051137

1
2
3
4
5
6
7
8

Diagnostic message
Symbol for event level
Status signal
Short text
Symbol for event level, status signal, diagnostic number
Remedial measureService ID
Operating time of occurrence

Symbol for event level
 "Alarm" status
Measurement is interrupted. The signal outputs adopt the defined alarm state. A
diagnostic message is generated.
 "Warning" status
The device continues to measure. A diagnostic message is generated.

11.4

Remedy informationCalling up

11.4.1Graphic display with keys

1→S

XXXXXXXXX

S801
Supply voltageiMenu

A0051131‣ Open the Diagnostic list submenu

54

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Diagnostics and troubleshooting

2→

Diagnostic list
Diagnostics 1

S

S801 Supply voltage

Diagnostics 2
Diagnostics 3

A0051132‣ Select the diagnostic event and confirm
3

Supply voltage(ID:203)
S801 0d00h02m25s
Increase supply voltage

A0051133‣ Remedy informationClosing

11.4.2

Operating menu

In the Diagnostic list submenu, up to 5 currently pending diagnostic messages can bedisplayed. If more than 5 messages are pending, the messages with the highest priority are
shown on the display.
Navigation path
Diagnostics → Diagnostic list
Queued diagnostic messages are also displayed in the Active diagnostics parameter.
Navigation: Diagnostics → Active diagnostics

11.5

Adapting the diagnostic information

The event level can be configured:
Navigation: Diagnostics → Diagnostic settings → Configuration

11.6

List of diagnostic events

If Remedy information Contact the Service Department is displayed(www.addresses.endress.com), be ready with the service ID shown.
Diagnostic
numberShort text

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

Diagnostic of sensor

Endress+Hauser

062

Sensor connection
faultyCheck sensor connection

151

Sensor electronic failure 1. Restart device
2. Contact service

F

Alarm

F

Alarm

55

Diagnostics and troubleshooting

Micropilot FMR60B PROFINET with Ethernet-APL

Diagnostic
number

168

Short text

Build-up detected

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

Check process conditions

M

Warning 1)

Diagnostic of electronic
232

Real time clock
defective

Replace main electronics

M

Warning

242

Firmware incompatible

1. Check software
2. Flash or change main electronic
module

F

Alarm

252

Module incompatible

1. Check if correct electronic
module is plugged
2. Replace electronic module

F

Alarm

270

Main electronics
defective

Replace main electronics

F

Alarm

272

Main electronics faulty

1. Restart device
2. Contact service

F

Alarm

273

Main electronics
defective

Replace main electronics

F

Alarm

282

Data storage
inconsistent

Restart device

F

Alarm

283

Memory content
inconsistent

1. Restart device
2. Contact service

F

Alarm

287

Memory content
inconsistent

1. Restart device
2. Contact service

M

Warning

331

Firmware update failed

1. Update firmware of device
2. Restart device

M

Warning388

Electronics andHistoROM defective

1. Restart device
2. Replace electronics andHistoROM
3. Contact service

F

Alarm

Diagnostic of configuration

56

410

Data transfer failed

1. Retry data transfer
2. Check connection

F

Alarm

412

Processing download

Download active, please waitCWarning435

Linearization faulty

Check linearization table

F

Alarm

436

Date/time incorrect

Check date and time settings.

M

Warning 1)

437

Configuration
incompatible

1. Update firmware
2. Execute factory reset

F

Alarm

438

Dataset different

1. Check dataset file
2. Check device parameterization
3. Download new device
parameterization

M

Warning

484

Failure mode
simulation active

Deactivate simulation

C

Alarm

485

Process variable
simulation active

Deactivate simulationC

Warning495

Diagnostic event
simulation active

Deactivate simulation

S

Warning538

Configuration Sensor
Unit invalid

1. Check sensor configuration
2. Check device configuration

F

Alarm

585

Simulation distance

Deactivate simulationC

WarningEndress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Diagnostic
number586

Diagnostics and troubleshooting

Short text

Record map

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

Recording of mapping
please waitCWarningDiagnostic of process

1)

801

Supply voltage too low

Increase supply voltage

S

Warning802

Supply voltage too highDecrease supply voltage

S

Warning825Electronics temperature 1. Check ambient temperature
2. Check process temperatureS

Warning826

Sensor temperature outof range

1. Check ambient temperature
2. Check process temperatureS

Warning

941

Echo lost

Check parameter 'DC value'S

Warning 1)

942

In safety distance

1. Check level
2. Check safety distance
3. Reset self holding

S

Warning 1)

952

Foam detected

Check process conditions

S

Warning 1)

968

Level limited

1. Check level
2. Check limit parameters

S

WarningDiagnostic behavior can be changed.

11.7

Event logbook

11.7.1

Event history

A chronological overview of the event messages that have occurred is provided in the
Event logbook submenu. This submenu only exists if operating via the local display withkeys. In the case of operation via FieldCare, the event list can be displayed with the "Event
List / HistoROM" functionality of FieldCare.
Navigation:
Diagnostics → Event logbook → Event list
A maximum of 100 event messages can be displayed in chronological order.
The event history includes entries for:• Diagnostic events• Information events
In addition to the operating time when the event occurred, each event is also assigned asymbol that indicates whether the event has occurred or is finished:• Diagnostic event• : Occurrence of the event• : End of the event• Information event
: Occurrence of the event

Calling up and closing the remedial measures

Endress+Hauser

1.

Press .
 The message for the remedial measures for the selected diagnostic event opens.

2.

Press  +  simultaneously.
 The message about the remedial measures closes.

57

Diagnostics and troubleshooting

Micropilot FMR60B PROFINET with Ethernet-APL

11.7.2Filtering the event logbook

Filters can be used to determine which category of event messages is displayed in the
Event list submenu.
Navigation: Diagnostics → Event logbook
Filter categories• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information

11.7.3

Overview of information events

Info number

58

Info name

I1000--------(Device ok)

I1079

Sensor changed

I1089

Power on

I1090

Configuration reset

I1091

Configuration changed

I11036

Date/time set successfully

I11074

Device verification active

I1110

Write protection switch changed

I1151

History reset

I1154

Reset terminal voltage min/max

I1155

Reset electronics temperature

I1157

Memory error event list

I1256

Display: access status changed

I1264

Safety sequence aborted

I1335

Firmware changed

I1397

Fieldbus: access status changed

I1398

CDI: access status changed

I1440

Main electronic module changed

I1444

Device verification passed

I1445

Device verification failed

I1461

Sensor verification failed

I1512

Download started

I1513

Download finished

I1514

Upload started

I1515

Upload finished

I1551

Assignment error fixed

I1552

Failed: Main electronic verification

I1554

Safety sequence started

I1555

Safety sequence confirmed

I1556

Safety mode off

I1663

Power off

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Diagnostics and troubleshooting

Info number

Info name

I1666

Clock synchronized

I1712

New flash file received

I1956

Reset

11.8

Resetting the device

11.8.1

Reset password via operating toolEnter a code to reset the current 'Maintenance' password.
The code is deliverd by your local support.
Navigation: System → User management → Reset password → Reset password
For detailed information on the Reset password parameter: Description of device
parameters.

11.8.2

Reset device via operating tool

Reset the device configuration - either entirely or in part - to a defined state
Navigation: System → Device management → Reset device
For detailed information on the Reset device parameter: Description of device
parameters.

11.8.3

Resetting the device via keys on the electronic insert

Resetting the password
15 s
I

I

I

I

Display

off onI

2×
A0050210

 21

Sequence for resetting the password

Delete/reset the password
1.

Press operating key I three times.
 The reset password function is started, the LED flashes.

2.

Press operating key I once within 15 s.
 The password is reset, the LED flashes briefly.

If operating key I is not pressed within 15 s, the action is canceled and the LED is no longer
lit.

Endress+Hauser

59Diagnostics and troubleshooting

Micropilot FMR60B PROFINET with Ethernet-APL

Resetting the device to the factory setting
12 s

off onI
DisplayI

50 ms
A0050015

 22

Sequence - Factory setting

Reset the device to the factory setting‣ Press operating key I for at least 12 s.
 Device data are reset to the factory setting, the LED flashes briefly.

11.9

Device information

All device information is contained in the Information submenu.
Navigation: System → Information
For detailed information on the Information submenu: Description of device
parameters.

11.10 Firmware history
The firmware version can explicitly be ordered via the product structure. This makes it
possible to ensure the compatibility of the firmware version with an existing orplanned system integration.
Version01.00.00• Initial software• Valid from: 01.05.2023

60

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

12

Maintenance

MaintenanceNo special maintenance work is required.

12.1

Exterior cleaning

Notes on cleaning• The cleaning agents used should not corrode the surfaces and the seals• Observe the degree of protection of the device

12.2

Seals

The process seals, located on the process connection of the device, should be replaced
periodically. The interval between changes depends on the frequency of the cleaning
cycles, the cleaning temperature and the medium temperature.

Endress+Hauser

61

Repair

Micropilot FMR60B PROFINET with Ethernet-APL

13

Repair

13.1

General information

13.1.1

Repair concept

Endress+Hauser repair concept• The devices have a modular design• Repairs can be carried out by Endress+Hauser Service or by appropriately trained
customers• Spare parts are grouped into logical kits with the associated replacement instructions
For more information on service and spare parts, please contact your Endress+Hauser
sales representative.

13.1.2

Repair of Ex-certified devices

LWARNING
Incorrect repair can compromise electrical safety!
Risk of explosion!
‣ Only specialist personnel or the Endress+Hauser service team may carry out repairs on
Ex-certified devices in accordance with national regulations.
‣ Relevant standards and national regulations on hazardous areas, safety instructions
and certificates must be observed.
‣ Use only original Endress+Hauser spare parts.
‣ Please note the device designation on the nameplate. Only identical parts may be used
as replacements.
‣ Carry out repairs according to the instructions.
‣ Only the Endress+Hauser service team is permitted to modify a certified device andconvert it to another certified version.

13.2

Spare parts• Some replaceable device components are identified by a spare part nameplate. This
contains information about the spare part.
• All the spare parts for the measuring device, along with the order code, are listed in the
Device Viewer (www.endress.com/deviceviewer) and can be ordered. If available, users
can also download the associated Installation Instructions.
Device serial number or QR code:
Located on the device and spare part nameplate.

13.3

Replacement

LCAUTION
Data upload/download is not permitted if the device is used for safety-related
applications.
‣ After an entire device or an electronics module has been replaced, the parameters canbe downloaded to the device again via the communication interface. For this, the data
must have been uploaded to the PC beforehand using the "FieldCare/DeviceCare"software.

62

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

13.3.1

RepairHistoROM

It is not necessary to perform a new device calibration after replacing the display ortransmitter electronics.
The spare part is supplied without HistoROM.
After removing the transmitter electronics, remove HistoRom and insert it into the
new spare part.

13.4

ReturnThe requirements for safe device return can vary depending on the device type andnational legislation.
1.

Refer to the web page for information:
http://www.endress.com/support/return-material
 Select the region.

2.

Return the device if repairs or a factory calibration are required, or if the wrong
device was ordered or delivered.

13.5

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize thedisposal of WEEE as unsorted municipal waste. Do not dispose of products bearing thismarking as unsorted municipal waste. Instead, return them to the manufacturer for
disposal under the applicable conditions.

Endress+Hauser

63

Accessories

Micropilot FMR60B PROFINET with Ethernet-APL

14

Accessories

14.1

Weather protection cover 316L

The weather protection cover can be ordered together with the device via the "Accessory
enclosed" product structure.
It is used to protect against direct sunlight, precipitation and ice.
Weather protection cover 316L is suitable for the dual compartment housing made ofaluminum or 316L. The delivery includes the holder for direct mounting on the housing.
228.9 (9.01)
136.4 (5.37)
92.5 (3.64)

135 (5.31)

102.5 (4.04)

73.5 (2.89)

170.5 (6.71)!65 (2.5
6)
81 (3.19)
103 (4.06)
A0039231

 23

Dimensions. Unit of measurement mm (in)Material• Weather protection cover: 316L
• Clamping screw: A4• Holder: 316L
Order number for accessories:
71438303

14.2

Plastic weather protection cover

The weather protection cover can be ordered together with the device via the "Accessory
enclosed" product structure.
It is used to protect against direct sunlight, precipitation and ice.
The plastic weather protection cover is suitable for the single compartment housing madeof aluminum. The delivery includes the holder for direct mounting on the housing.

122 (4.8)!115 (4.53)

140 (5.51)
165 (6.5)

32 (1.26)
140 (5.51)
A0038280

 24

64

Dimensions. Unit of measurement mm (in)Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Accessories

Material
Plastic
Order number for accessories:
71438291

14.3

Mounting bracket, adjustable

The device can be mounted on a wall or ceiling with the mounting bracket.
The device can be aligned with the product surface with the swivel function.
The mounting bracket can be ordered together with the device via the product structure"Accessory enclosed".
Suitable for device with single compartment housing or dual compartment aluminum
housing, L-form, in combination with encapsulated antenna, PVDF, 40 mm (1.5 in) or
drip-off antenna 50 mm (2 in) with threaded process connection.

A0048745

 25

Ceiling or wall mounting

There is no conductive connection between the mounting bracket and the transmitter
housing. The bracket should be included in local potential equalization to prevent any
electrostatic charge.
Fasten only to stable materials (e.g. metal, brick, concrete) with suitable fastening
fixtures (provided by the customer).
Order number for accessories:
71597288

14.3.1

Dimensions

3 (0.12)9 (0.35)!9 (0.35)

35 (1.38)

6 (0.24)156 (6.14)

65 (2.56)
32.5 (1.28)

282 (11.1)
230 (9.06)

A0048769

 26

Endress+Hauser

Dimensions of mounting bracket. Unit of measurement mm (in)65

Accessories

Micropilot FMR60B PROFINET with Ethernet-APL

14.3.2

Scope of delivery

5
A0049050

 27

Scope of delivery of mounting bracket, adjustable

1 × mounting bracket, 316L (1.4404)
2 × holder, 316L (1.4404)
6 × screws, A4
4 × lock washer, A4

14.4

M12 socket

A0051231

 28

M12 socket, straightM12 socket, straight• Material:
Body: PBT; union nut: nickel-plated die-cast zinc; seal: NBR• Degree of protection (fully locked): IP67
• Pg coupling: Pg7• Order number: 52006263

A0051232

 29

M12 socket, angled

M12 socket, angled• Material:
Body: PBT; union nut: nickel-plated die-cast zinc; seal: NBR• Degree of protection (fully locked): IP67
• Pg coupling: Pg7• Order number: 71114212

66

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Accessories

A0051233

 30

M12 socket, angled, cableM12 socket, angled, 5 m (16 ft) cable• M12 socket material:• Body: TPU• Union nut: nickel-plated die-cast zinc
• Cable material:
PVC• Cable Li Y YM 4×0.34 mm2 (20 AWG)
• Cable colors• 1 = BN = brown• 2 = WH = white• 3 = BU = blue
• 4 = BK = black• Order number: 52010285

14.5

Remote display FHX50B

The remote display is ordered via the Product Configurator.
If the remote display is to be used, the device version Prepared for display FHX50B must
be ordered.

Endress+Hauser

67

Accessories

Micropilot FMR60B PROFINET with Ethernet-APL

A

D

E
B

F
C

G

A0046692

A
B
C
D
E
F
G

Plastic single compartment housing, remote display
Aluminum single compartment housing, remote display
Single compartment housing, 316L hygiene, remote display
Device side, plastic single compartment housing prepared for display FHX50B
Device side, aluminum single compartment housing prepared for display FHX50B
Device side, dual compartment housing, L-form, prepared for display FHX50B
Device side, single compartment housing, 316L hygiene, prepared for display FHX50B

Material of single compartment housing, remote display• Aluminum• Plastic
Degree of protection:• IP68 / NEMA 6P• IP66 / NEMA 4x
Connecting cable:• Connecting cable (option) up to 30 m (98 ft)• Customer-supplied standard cable up to 60 m (197 ft)
Recommendation: EtherLine®-P CAT.5e from LAPP.
Specification of customer-supplied connecting cablePush-in CAGE CLAMP®, connection technology, push actuation• Conductor cross-section:• Solid conductor 0.2 to 0.75 mm2 (24 to 18 AWG)
• Fine-stranded conductor 0.2 to 0.75 mm2 (24 to 18 AWG)
• Fine-stranded conductor; with insulated ferrule 0.25 to 0.34 mm2• Fine-stranded conductor; without insulated ferrule 0.25 to 0.34 mm2• Stripping length 7 to 9 mm (0.28 to 0.35 in)• Outer diameter: 6 to 10 mm (0.24 to 0.4 in)• Maximum cable length: 60 m (197 ft)
Ambient temperature:• –40 to +80 °C (–40 to +176 °F)• Option: –50 to +80 °C (–58 to +176 °F)

68Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

14.6

Accessories

Gas-tight feedthroughChemically inert glass feedthrough, which prevents gases from entering the electronics
housing.
Can optionally be ordered as "Accessory mounted" via the product structure.

14.7

Field Xpert SMT70

Universal, high-performance tablet PC for device configuration in Ex Zone 2 and non-Ex
areas
For details, see "Technical Information" TI01342S

14.8

DeviceCare SFE100

Configuration tool for HART, PROFIBUS and FOUNDATION Fieldbus field devices
Technical Information TI01134S

14.9

FieldCare SFE500

FDT-based plant asset management tool
It can configure all smart field units in your system and helps you manage them. By using
the status information, it is also a simple but effective way of checking their status andcondition.
Technical Information TI00028S

Endress+Hauser

69

Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

15

Technical data

15.1

Input

Measured variable

The measured variable is the distance between the reference point and the productsurface. The level is calculated based on "E", the empty distance entered.

Measuring range

The measuring range starts at the point where the beam hits the tank floor. Levels below
this point cannot be measured, particularly in the case of spherical bases or conical outlets.
Maximum measuring range
The maximum measuring range depends on the antenna size and design.
Antenna

Maximum measuring range

Encapsulated, PVDF, 40 mm (1.5 in)

40 m (131 ft)

Drip-off, PTFE, 50 mm (2 in)

50 m (164 ft)

Integrated, PEEK, 20 mm (0.75 in)

10 m (32.8 ft)

Integrated, PEEK, 40 mm (1.5 in)

22 m (72 ft)

Usable measuring range
The usable measuring range depends on the antenna size, the medium's reflective
properties, the installation position and any possible interference reflections.
In principle, measurement is possible up to the tip of the antenna.
To avoid any material damage from corrosive or aggressive media or deposit buildup on
the antenna, the end of the measuring range should be selected 10 mm (0.4 in) before thetip of the antenna.

70

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

A

R

0%

C

B

H

100%

A0051658

 31
A
B
C
H
R

Usable measuring range

Length of antenna + 10 mm (0.4 in)
Usable measuring range
50 to 80 mm (1.97 to 3.15 in); medium εr < 2
Vessel height
Reference point of measurement, varies depending on the antenna system

For further information on the reference point, see →A

Mechanical construction.

In the case of media with a low dielectric constant, εr < 2 , the tank floor may be visible
through the medium at very low levels (lower than level C). Reduced accuracy must be
expected in this range. If this is not acceptable, the zero point should be positioned at a
distance C above the tank floor in these applications → Usable measuring range.
The media groups and the possible measuring range are described as a function of theapplication and media group in the following section. If the dielectric constant of themedium is not known, to ensure a reliable measurement assume the medium belongs togroup B.

Endress+Hauser

71Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

Media groups• A0 (εr 1.2 to 1.4)
e.g. n-butane, liquid nitrogen, liquid hydrogen• A (εr 1.4 to 1.9)
Non-conductive liquids, e.g. liquefied gas• B (εr 1.9 to 4)
Non-conductive liquids, e.g. gasoline, oil, toluene, etc.
• C (εr 4 to 10)
e.g. concentrated acid, organic solvents, ester, aniline, etc.
• D (εr >10)
Conductive liquids, aqueous solutions, diluted acids, bases and alcohol
Measurement of the following media with absorbing gas phase
For example:• Ammonia• Acetone• Methylene chloride• Methyl ethyl ketone• Propylene oxide• VCM (vinyl chloride monomer)
To measure absorbing gases, either use a guided radar, measuring devices withanother measuring frequency or another measuring principle.
If measurements must be performed in one of these media, please contact Endress+Hauser.
For the dielectric constants (DC values) of many media commonly used in industry,please refer to:• Dielectric constant (DC value) Compendium CP01076F• The Endress+Hauser "DC Values app" (available for Android and iOS)
Measurement in storage vessel
Storage vessel - measuring conditions
Calm medium surface (e.g. bottom filling, filling via immersion tube or rare filling fromabove)
Integrated antenna, PEEK, 20 mm (0.75 in) in storage vessel

72

Media group

Measuring range

A0 (εr 1.2 to 1.4)

1.5 m (5 ft)

A (εr 1.4 to 1.9)

2.5 m (8 ft)

B (εr 1.9 to 4)

5 m (16 ft)

C (εr 4 to 10)

8 m (26 ft)

D (εr >10)

10 m (33 ft)

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

Integrated antenna, PEEK, 40 mm (1.5 in) in storage vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

3 m (10 ft)

A (εr 1.4 to 1.9)

6 m (20 ft)

B (εr 1.9 to 4)

11 m (36 ft)

C (εr 4 to 10)

15 m (49 ft)

D (εr >10)

22 m (72 ft)

Encapsulated antenna, PVDF, 40 mm (1.5 in) in storage vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

7 m (23 ft)

A (εr 1.4 to 1.9)

15 m (49.2 ft)

B (εr 1.9 to 4)

30 m (98.4 ft)

C (εr 4 to 10)

40 m (131 ft)

D (εr >10)40 m (131 ft)

PTFE Drip-off antenna, 50 mm (2 in) in storage vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

7 m (23 ft)

A (εr 1.4 to 1.9)

12 m (39 ft)

B (εr 1.9 to 4)

23 m (75 ft)

C (εr 4 to 10)

40 m (131 ft)

D (εr >10)50 m (164 ft)

Measurement in buffer vessel
Buffer vessel - measuring conditions
Moving medium surface (e.g. permanent free filling from above, mixing jets)

Endress+Hauser

73

Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

Integrated antenna, PEEK, 40 mm (1.5 in) in buffer vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)

1.5 m (5 ft)

A (εr 1.4 to 1.9)

3 m (10 ft)

B (εr 1.9 to 4)

6 m (20 ft)

C (εr 4 to 10)

13 m (43 ft)

D (εr >10)20 m (66 ft)

Encapsulated antenna, PVDF, 40 mm (1.5 in) in buffer vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)4 m (13 ft)

A (εr 1.4 to 1.9)

7.5 m (24.6 ft)

B (εr 1.9 to 4)

15 m (49.2 ft)

C (εr 4 to 10)

25 m (82 ft)

D (εr >10)35 m (114.8 ft)

PTFE Drip-off antenna, 50 mm (2 in) in buffer vessel
Media group

Measuring range

A0 (εr 1.2 to 1.4)4 m (13 ft)

A (εr 1.4 to 1.9)7 m (23 ft)

B (εr 1.9 to 4)

13 m (43 ft)

C (εr 4 to 10)

28 m (92 ft)

D (εr >10)44 m (144 ft)

Measurement in vessel with agitator
Vessel with agitator - measuring conditions
Turbulent medium surface (e.g. from filling from above, stirrers and baffles)

74

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

Integrated antenna, PEEK, 20 mm (0.75 in) in vessel with agitator
Media group

Measuring range

A (εr 1.4 to 1.9)

1 m (3.3 ft)

B (εr 1.9 to 4)

1.5 m (5 ft)

C (εr 4 to 10)

3 m (10 ft)

D (εr >10)

5 m (16 ft)

Integrated antenna, PEEK, 40 mm (1.5 in) in vessel with agitator
Media group

Measuring range

A0 (εr 1.2 to 1.4)

1 m (3.3 ft)

A (εr 1.4 to 1.9)

1.5 m (5 ft)

B (εr 1.9 to 4)

3 m (10 ft)

C (εr 4 to 10)

7 m (23 ft)

D (εr >10)

11 m (36 ft)

Encapsulated antenna, PVDF, 40 mm (1.5 in) in vessel with agitator

Endress+Hauser

Media group

Measuring range

A0 (εr 1.2 to 1.4)

2 m (7 ft)

A (εr 1.4 to 1.9)

4 m (13 ft)

B (εr 1.9 to 4)

5 m (16.4 ft)

C (εr 4 to 10)

15 m (49.2 ft)

D (εr >10)20 m (65.6 ft)

75

Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

PTFE Drip-off antenna, 50 mm (2 in) in vessel with agitator

Operating frequency

Media group

Measuring range

A0 (εr 1.2 to 1.4)

2 m (7 ft)

A (εr 1.4 to 1.9)

4 m (13 ft)

B (εr 1.9 to 4)

7 m (23 ft)

C (εr 4 to 10)

15 m (49 ft)

D (εr >10)

25 m (82 ft)

Approx. 80 GHz
Up to 8 devices can be installed in a tank without the devices mutually influencing one
another.

Transmission power• Peak power: 6.3 mW• Average output power: 63 µW

15.2

OutputPROFINET-APL

PROFINET with Ethernet-APL
10BASE-T1L, 2-wire 10 Mbit/s

Signal on alarm

Local display
Status signal (in accordance with NAMUR Recommendation NE 107):
Plain text display
Operating tool via service interface (CDI)
Status signal (in accordance with NAMUR Recommendation NE 107):
Plain text display
Operating tool via PROFINET with Ethernet-APL• According to "Application layer protocol for decentralized periphery", Version 2.4• Diagnostics according to PROFINET PA Profile 4.02

Linearization

The linearization function of the device allows the conversion of the measured value into
any unit of length, weight, flow or volume.
Pre-programmed linearization curves
Linearization tables for calculating the volume in the following vessels are preprogrammed
into the device:• Pyramid bottom• Conical bottom• Angled bottom• Horizontal cylinder• Sphere
Other linearization tables of up to 32 value pairs can be entered manually.

76

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

PROFINET with EthernetAPL

Technical data

ProtocolApplication layer protocol for decentral device periphery and distributed
automation, Version 2.4

Communication type

Ethernet Advanced Physical Layer 10BASE-T1L

Conformance Class

Conformance Class B

Netload Class

Netload Class II

Baud rates

Automatic 10 Mbit/s with full-duplex detection

Cycle times

From 32 ms

Polarity

Auto-polarity for automatic correction of crossed TxD and RxD pairs

Media Redundancy Protocol
(MRP)

Yes

System redundancy support

System redundancy S2 (2 AR with 1 NAP)

Device profile

Application interface identifier 0xB321
Generic device

Manufacturer ID

0x11

Device type ID

0xA1C1

Device description files (GSD,
FDI, DTM, DD)

Information and files available at:• www.endress.com
On the product page for the device: Documents/Software → Device drivers• www.profibus.org

Supported connections• 2 x AR (IO Controller AR)• 1 x AR (IO-Supervisor Device AR connection allowed)• 1 x Input CR (Communication Relation)• 1 x Output CR (Communication Relation)• 1 x Alarm CR (Communication Relation)

Configuration options fordevice• Manufacturer-specific software (FieldCare, DeviceCare)• Web browser• Device master file (GSD), can be read out via the integrated Web server ofthe device• DIP switch for setting the service IP address

Configuration of thedevice name

• DCP protocol• Process Device Manager (PDM)• Integrated Web server

Supported functions• Identification & maintenance
Simple device identification via:
• Control system• Nameplate• Measured value status
The process variables are communicated with a measured value status• Blinking feature via the local display for simple device identification andassignment• Device operation via operating tools (e.g., FieldCare, DeviceCare, SIMATIC
PDM)

System integration

For information on system integration, see

Operating Instructions• Cyclic data transmission• Overview and description of the modules• Status coding• Startup configuration• Factory setting

Endress+Hauser

77Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

15.3
Ambient temperature
range

Environment

The following values apply up to a process temperature of +85 °C (+185 °F). At higher
process temperatures, the permitted ambient temperature is reduced.
• Without LCD display:Standard: –40 to +85 °C (–40 to +185 °F)• With LCD display: –40 to +85 °C (–40 to +185 °F) with limitations in optical properties
such as display speed and contrast for example. Can be used without limitations up to–20 to +60 °C (–4 to +140 °F)
If operating outdoors in strong sunlight:• Mount the device in the shade.
• Avoid direct sunlight, particularly in warm climatic regions.
• Use a weather protection cover (see accessories).

Ambient temperature limits

The permitted ambient temperature (Ta) depends on the selected housing material(Product Configurator → Housing; Material →) and the selected process temperature range(Product Configurator → Application →).
In the event of temperature (Tp) at the process connection, the permitted ambient
temperature (Ta) is reduced.
The following information only takes functional aspects into consideration. Additional
restrictions may apply for certified device versions.
Plastic housing
Plastic housing; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 32
P1
P2
P3
P4
P5

Plastic housing; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –20 to +150 °C (–4 to +302 °F) is limited to
0 to +150 °C (+32 to +302 °F).

78

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

Restriction to a process temperature of 0 to +150 °C (+32 to +302 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 33
P1
P2
P3
P4
P5

Plastic housing; process temperature 0 to +150 °C (+32 to +302 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Plastic housing; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 34
P1
P2
P3
P4
P5

Plastic housing; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –20 to +200 °C (–4 to +392 °F) is limited to
0 to +200 °C (+32 to +392 °F).
Restriction to a process temperature of 0 to +200 °C (+32 to +392 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 35
P1
P2
P3
P4
P5

Endress+Hauser

Plastic housing; process temperature 0 to +200 °C (+32 to +392 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

79

Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

Plastic housing; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 36
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +80 °C (+176 °F) | Ta: +75 °C (+167 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of–40 to +80 °C (–40 to +176 °F) is limited to 0 to +80 °C (+32 to +176 °F).
Restriction to a process temperature of 0 to +80 °C (+32 to +176 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 37
P1
P2
P3
P4
P5

80

Plastic housing; process temperature 0 to +80 °C (+32 to +176 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +80 °C (+176 °F) | Ta: +75 °C (+167 °F)= Tp: +80 °C (+176 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

Plastic housing; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 38
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +130 °C (+266 °F) | Ta: +41 °C (+106 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –40 to +130 °C (–40 to +266 °F) is limited to
0 to +130 °C (+32 to +266 °F).
Restriction to a process temperature of 0 to +130 °C (+32 to +266 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 39
P1
P2
P3
P4
P5

Plastic housing; process temperature 0 to +130 °C (+32 to +266 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +130 °C (+266 °F) | Ta: +41 °C (+106 °F)= Tp: +130 °C (+266 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Plastic housing; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 40
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –40 to +150 °C (–40 to +302 °F) is limited to
0 to +150 °C (+32 to +302 °F).
Endress+Hauser

81Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

Restriction to a process temperature of 0 to +150 °C (+32 to +302 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 41
P1
P2
P3
P4
P5

Plastic housing; process temperature 0 to +150 °C (+32 to +302 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +25 °C (+77 °F)= Tp: +150 °C (+302 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Plastic housing; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 42
P1
P2
P3
P4
P5

Plastic housing; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

In the case of devices with a plastic housing and CSA C/US approval, the selected
process temperature of –40 to +200 °C (–40 to +392 °F) is limited to
0 to +200 °C (+32 to +392 °F).
Restriction to a process temperature of 0 to +200 °C (+32 to +392 °F) with CSA C/US
approval and plastic housing
Ta
P1

P2
P3

P5

P4

Tp
A0048826

 43
P1
P2
P3
P4
P5

82

Plastic housing; process temperature 0 to +200 °C (+32 to +392 °F) with CSA C/US approval= Tp: 0 °C (+32 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +27 °C (+81 °F)= Tp: +200 °C (+392 °F) | Ta: 0 °C (+32 °F)= Tp: 0 °C (+32 °F) | Ta: 0 °C (+32 °F)

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

Aluminum housing, coated
Aluminum housing; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 44
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +150 °C (+302 °F) | Ta: +53 °C (+127 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

Aluminum housing; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 45
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +200 °C (+392 °F) | Ta: +47 °C (+117 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

Aluminum housing; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 46
P1
P2
P3
P4
P5

Endress+Hauser

Aluminum housing, coated; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +80 °C (+176 °F) | Ta: +79 °C (+174 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

83Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

Aluminum housing; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 47
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +130 °C (+266 °F) | Ta: +55 °C (+131 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Aluminum housing; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 48
P1
P2
P3
P4
P5

Aluminum housing, coated; process temperature –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +150 °C (+302 °F) | Ta: +53 °C (+127 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Aluminum housing; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 49
P1
P2
P3
P4
P5

84

Aluminum housing, coated; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +79 °C (+174 °F)= Tp: +79 °C (+174 °F) | Ta: +79 °C (+174 °F)= Tp: +200 °C (+392 °F) | Ta: +47 °C (+117 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

316L housing
316L housing; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 50
P1
P2
P3
P4
P5

316L housing; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +150 °C (+302 °F) | Ta: +43 °C (+109 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 51
P1
P2
P3
P4
P5

316L housing; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +200 °C (+392 °F) | Ta: +38 °C (+100 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 52
P1
P2
P3
P4
P5

Endress+Hauser

316L housing; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +80 °C (+176 °F) | Ta: +77 °C (+171 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

85Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

316L housing; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 53
P1
P2
P3
P4
P5

316L housing; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +130 °C (+266 °F) | Ta: +54 °C (+129 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 54
P1
P2
P3
P4
P5

316L housing; process temperature range: –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +150 °C (+302 °F) | Ta: +43 °C (+109 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 55
P1
P2
P3
P4
P5

86

316L housing; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +77 °C (+171 °F)= Tp: +77 °C (+171 °F) | Ta: +77 °C (+171 °F)= Tp: +200 °C (+392 °F) | Ta: +38 °C (+100 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

316L housing, hygiene
316L housing, hygiene; process temperature –20 to +150 °C (–4 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 56
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –20 to +150 °C (–4 to +302 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +41 °C (+106 °F)= Tp: +150 °C (+302 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing, hygiene; process temperature –20 to +200 °C (–4 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 57
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –20 to +200 °C (–4 to +392 °F)= Tp: –20 °C (–4 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +32 °C (+90 °F)= Tp: +200 °C (+392 °F) | Ta: –20 °C (–4 °F)= Tp: –20 °C (–4 °F) | Ta: –20 °C (–4 °F)

316L housing, hygiene; process temperature –40 to +80 °C (–40 to +176 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 58
P1
P2
P3
P4
P5

Endress+Hauser

316L housing, hygiene; process temperature –40 to +80 °C (–40 to +176 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +80 °C (+176 °F) | Ta: +75 °C (+167 °F)= Tp: +80 °C (+176 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

87Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

316L housing, hygiene; process temperature –40 to +130 °C (–40 to +266 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 59
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –40 to +130 °C (–40 to +266 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +130 °C (+266 °F) | Ta: +55 °C (+131 °F)= Tp: +130 °C (+266 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing, hygiene; process temperature –40 to +150 °C (–40 to +302 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 60
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature range: –40 to +150 °C (–40 to +302 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +150 °C (+302 °F) | Ta: +41 °C (+106 °F)= Tp: +150 °C (+302 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

316L housing, hygiene; process temperature –40 to +200 °C (–40 to +392 °F)
Ta
P1

P2
P3

P5

P4

Tp
A0032024

 61
P1
P2
P3
P4
P5

316L housing, hygiene; process temperature –40 to +200 °C (–40 to +392 °F)= Tp: –40 °C (–40 °F) | Ta: +76 °C (+169 °F)= Tp: +76 °C (+169 °F) | Ta: +76 °C (+169 °F)= Tp: +200 °C (+392 °F) | Ta: +32 °C (+90 °F)= Tp: +200 °C (+392 °F) | Ta: –40 °C (–40 °F)= Tp: –40 °C (–40 °F) | Ta: –40 °C (–40 °F)

Storage temperature• Without LCD display: –40 to +90 °C (–40 to +194 °F)• With LCD display: –40 to +85 °C (–40 to +185 °F)

Climate class

DIN EN 60068-2-38 (test Z/AD)

88

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Installation height as per
IEC61010-1 Ed.3

Generally up to 5 000 m (16 404 ft) above sea level

Degree of protection

Test as per IEC 60529 and NEMA 250-2014

Technical data

Housing
IP66/68, NEMA TYPE 4X/6P
IP68 test condition: 1.83 m under water for 24 hours.
Cable entries• Gland M20, plastic, IP66/68 NEMA TYPE 4X/6P• Gland M20, nickel-plated brass, IP66/68 NEMA TYPE 4X/6P• Gland M20, 316L, IP66/68 NEMA TYPE 4X/6P• Gland M20, hygiene, IP66/68/69 NEMA Type 4X/6P• Thread M20, IP66/68 NEMA TYPE 4X/6P• Thread G1/2, IP66/68 NEMA TYPE 4X/6P
If the G1/2 thread is selected, the device is delivered with an M20 thread as standard
and a G1/2 adapter is included with the delivery, along with the corresponding
documentation• Thread NPT1/2, IP66/68 NEMA TYPE 4X/6P• M12 plug
• When housing is closed and connecting cable is plugged in: IP66/67 NEMA TYPE 4X• When housing is open or connecting cable is not plugged in: IP20, NEMA TYPE 1
NOTICEM12 plug: Loss of IP protection class due to incorrect installation!
‣ The degree of protection only applies if the connecting cable used is plugged in andscrewed tight.
‣ The degree of protection only applies if the connecting cable used is specified according
to IP66/67 NEMA 4X.
‣ The protection classes are only maintained if the dummy cap is used or the cable isconnected.
Vibration resistanceDIN EN 60068-2-64 / IEC 60068-2-64 for 5 to 2 000 Hz: 1.5 (m/s2)2/Hz

Electromagnetic
compatibility (EMC)• Electromagnetic compatibility as per EN 61326 series and NAMUR recommendation
EMC (NE21)• Maximum measured error during EMC testing: < 0.5 % of the current digital measured
value
For more details refer to the EU Declaration of Conformity.

Endress+Hauser

89Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

15.4
Process pressure range

Process

LWARNING
The maximum pressure for the device depends on the lowest-rated component withregard to pressure (components are: process connection, optional mounted parts oraccessories).
‣ Only operate the device within the specified limits for the components!
‣ MWP (Maximum Working Pressure): The MWP is specified on the nameplate. Thisvalue refers to a reference temperature of +20 °C (+68 °F) and may be applied to thedevice for an unlimited time. Note temperature dependence of MWP. For flanges, referto the following standards for the permitted pressure values at higher temperatures:
EN 1092-1 (with regard to their stability/temperature property, the materials 1.4435
and 1.4404 are grouped together under EN 1092-1; the chemical composition of thetwo materials can be identical), ASME B16.5, JIS B2220 (the latest version of thestandard applies in each case). MWP data that deviate from this are provided in therelevant sections of the Technical Information.
‣ The Pressure Equipment Directive (2014/68/EU) uses the abbreviation PS . This
corresponds to the maximum working pressure (MWP) of the device.
The following tables show the dependencies between the seal material, processtemperature (TP) and process pressure range for each process connection that can beselected for the antenna used.
Encapsulated antenna, PVDF, 40 mm (1.5 in)Process connection thread 1-½"
Seal

Tp

Process pressure range

PVDF encapsulated–40 to +80 °C(–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

PVDF encapsulated–40 to +130 °C(–40 to +266 °F)–1 to 3 bar (–14.5 to 43.5 psi)

The following temperature restriction applies for devices with the dust ignition-proof
approval category 1D, 2D or 3D
A0047831

PVDF encapsulated–20 to +80 °C (–4 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

Seal

Tp

Process pressure range

PVDF encapsulated–40 to +80 °C(–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

Process connection UNI flange PP

The following temperature restriction applies for devices with the dust ignition-proof
approval category 1D, 2D or 3D
PVDF encapsulated–20 to +80 °C (–4 to +176 °F) –1 to 3 bar (–14.5 to 43.5 psi)

A0047947

The pressure range may be further restricted in the event of a CRN approval.

90

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Technical data

Drip-off antenna 50 mm (2 in)Process connection thread

A0047447

Seal

Tp

Process pressure range

FKM Viton GLT–40 to +130 °C (–40 to +266 °F)–1 to 16 bar (–14.5 to 232 psi)

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 16 bar (–14.5 to 232 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 16 bar (–14.5 to 232 psi)

EPDM

–40 to +130 °C (–40 to +266 °F)–1 to 16 bar (–14.5 to 232 psi)

HNBR

–20 to +150 °C (–4 to +302 °F)–1 to 16 bar (–14.5 to 232 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 16 bar (–14.5 to 232 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 16 bar (–14.5 to 232 psi)

Process connection UNI flange PP
Seal

Tp

Process pressure range

FKM Viton GLT–40 to +80 °C (–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

EPDM

–40 to +80 °C (–40 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

The following temperature restriction applies for devices with the HNBR or FFKM
Kalrez O-ring
A0047726

HNBR–20 to +80 °C (–4 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FFKM Kalrez–20 to +80 °C (–4 to +176 °F)–1 to 3 bar (–14.5 to 43.5 psi)

Process connection UNI flange 316L

A0047726

Seal

Tp

Process pressure range

FKM Viton GLT–40 to +130 °C (–40 to +266 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 3 bar (–14.5 to 43.5 psi)

EPDM

–40 to +130 °C (–40 to +266 °F)–1 to 3 bar (–14.5 to 43.5 psi)

HNBR

–20 to +150 °C (–4 to +302 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 3 bar (–14.5 to 43.5 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 3 bar (–14.5 to 43.5 psi)

The pressure range may be further restricted in the event of a CRN approval.
Integrated antenna, PEEK, 20 mm (0.75 in)Process connection thread 3/4"
Seal

Tp

Process pressure range

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

A0047832

The pressure range may be further restricted in the event of a CRN approval.

Endress+Hauser

91

Technical data

Micropilot FMR60B PROFINET with Ethernet-APL

Integrated antenna, PEEK, 40 mm (1.5 in)Process connection thread 1-½"
Seal

Tp

Process pressure range

FKM Viton GLT–40 to +150 °C (–40 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FKM Viton GLT–40 to +200 °C (–40 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +150 °C (–4 to +302 °F)–1 to 20 bar (–14.5 to 290 psi)

FFKM Kalrez–20 to +200 °C (–4 to +392 °F)–1 to 20 bar (–14.5 to 290 psi)

A0047833

The pressure range may be further restricted in the event of a CRN approval.

Dielectric constant

For liquidsεr ≥ 1.2
Contact Endress+Hauser for applications with lower dielectric constants than indicated.

92

Endress+Hauser

Micropilot FMR60B PROFINET with Ethernet-APL

Index

Index
Symbols"Device ID" parameter . . . . . . . . . . . . . . . . . . . . . . . . . 40
"Device revision" parameter . . . . . . . . . . . . . . . . . . . . 40
"Firmware version" parameter . . . . . . . . . . . . . . . . . . . 40
"Manufacturer ID" parameter . . . . . . . . . . . . . . . . . . . 40

A

Access authorization to parameters
Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Access code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Incorrect input . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

BBluetooth® wireless technology . . . . . . . . . . . . . . . . . 31

C

CE mark (Declaration of Conformity) . . . . . . . . . . . . . . 9
Cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 61
Climate class . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 88
Cyclic data transmission . . . . . . . . . . . . . . . . . . . . . . 42

D

Declaration of Conformity . . . . . . . . . . . . . . . . . . . . . . 9
Device description files . . . . . . . . . . . . . . . . . . . . . . . 40
Device locking, status . . . . . . . . . . . . . . . . . . . . . . . . 50
Device master file
GSD . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Device Viewer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Device description file . . . . . . . . . . . . . . . . . . . . . . 40
Diagnostic event . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Display values
For locking status . . . . . . . . . . . . . . . . . . . . . . . . . 50
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 63
Document
Purpose . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

EEvent history . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Event text . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . . 61

F

Field of application
Residual risks . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Device description file . . . . . . . . . . . . . . . . . . . . . . 40
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Filtering the event logbook . . . . . . . . . . . . . . . . . . . . 58
FirmwareRelease date . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40

I

MMedia . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

OOperation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9P

Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
PROFINET protocol . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Purpose of this document . . . . . . . . . . . . . . . . . . . . . . 5

R

Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Reading off measured values . . . . . . . . . . . . . . . . . . . 50
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . . . 6
Repair concept . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Requirements concerning the staff . . . . . . . . . . . . . . . . 8
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 63

S

Safety instructions
Basic . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Safety Instructions (XA) . . . . . . . . . . . . . . . . . . . . . . . . 6
Securing screw . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Service interface (CDI) . . . . . . . . . . . . . . . . . . . . . 39, 46
Settings
Adapting the measuring device to the process
conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Software release . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Nameplate . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Status signal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Submenu
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Interfaces . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 50
System redundancy S2 . . . . . . . . . . . . . . . . . . . . . . . . 44

T

Troubleshooting . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Turning the display module . . . . . . . . . . . . . . . . . . . . 19

U

Use of the measuring device
see Intended use
Use of the measuring devices
Borderline cases . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
Incorrect use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

VVersion data for the device . . . . . . . . . . . . . . . . . . . . 40

W

Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 30

Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

Endress+Hauser

93

*71610086*
71610086

www.addresses.endress.com