Wide flashing range 30–360,000 FPM
Brightness can be freely adjusted

LEDSTROBE
Check the motion
without stopping it.

Sturdy dustproof/waterproof body
Uniform illumination
16 hours of continuous operation
by a large-capacity battery

Easy-to-use! Bright! Long Life!
Sturdy IP54 dustproof/waterproof
portable LED stroboscope

Check for color, misalignments, faintness, omissions, andblemishes in printed matter
Applications Measuring the spindle rotational speed, yarn balloon inspection,and loom timing adjustment in textile production
Measure the motorspeed, and check rapidly movingmachinery parts
Surface inspection in steel and printing industries, andstudy and analysis of high-speed phenomena in R&D.

LED STROBE
Front rubber cover to mitigate external shocks

Carefully selected functions in a dustproof/waterproof sturdy bodyLEDs reduce running costs

Change settings such as flash
frequency with a knob

Targeted area is clearly visible
φ12cm

S-1 uniformly illuminates a range 12 cm in diameterfrom a distance of 30 cm.
Ultra-bright LEDs provide sufficient brightness.

30cm

Dustproof/waterproof sturdy bodySpecifications

IP54-compliant dustproof/waterproof structure
Usable in diverse workplaces without worrying about
dust, water droplets, etc.

Long-time cordless use
A large-capacity battery enables 16 hours of continuous
operation.
Can be used even at sites where no power is available.

Diverse functions such as delay,
frequency division, and slow motion
for controlling external trigger signal

Flash frequency range

Internal trigger: 30–360,000 FPM
External trigger: 0–36,000 FPM

Flash duration setting

Duty cycle(*): 0.1–1.0% (max 500 μs or less)
Time: 0.1–500.0 μs (1% or less of a flash cycle)

Operation modes

Internal trigger mode/External trigger mode/
External trigger divide mode

External trigger input signal Voltage signal, synchronized with rising edge
Low level: 0–1 V High level: 3–5 V

Memory function enables switchable
settings
Settings for up to 10 cases can be saved.
Settings can be easily changed even when using for
multiple flash conditions.

Diverse functions for controllingexternal trigger signal
Angle delay, time delay, frequency division, slow motion,etc. may be chosen according to the application.

Delay time

0.04 ms to the input signal cycle time or less，max 2000 ms (resolution 0.01 ms)

Delay angle

0 –359° (resolution 1°)

External signal frequency
division

Maximum division ratio: 1000

Slow motion-1.0 – +1.0 r/s (resolution 0.1 r/s)

Battery

Lithium-ion battery (built-in)
Battery life: 16 h (at maximum intensity, full charge)
Recharge time: 2 h (when power is OFF)IP performance

IP54 (Dustproof/waterproof)

AC adaptor

100–240 VAC±10%, 50–60 Hz

Weight

1.7 kgDimensions

L 223 × W 126 × H 164.5 (mm)(*) Duty cycle is the ratio of flash duration to flash cycle.

Read the instruction manual before using the device to ensure safety and to avoid damage to the device.

Can be used during charging

Head office
Tokyo sales office

8-2 Minami-Kurokawa, Asao-ku, Kawasaki-shi, Kanagawa, 215-0034, Japan
Tel: +81-44-989-7320 Fax: +81-44-989-7338

Osaka sales office6-17 Yokomakura-Nishi, Higashiosaka-shi, Osaka, 578-0956, Japan
Tel: +81-72-966-1061 Fax: +81-72-966-0961

Nagoya sales office 1-2-29 Kamimaezu, Naka-ku, Nagoya-shi, Aichi, 460-0013, Japan
Tel: +81-52-331-6562 Fax: +81-52-331-6604
URL: https://www.sugawara-labs.co.jp/

E-mail: info@sugawara-labs.co.jp●Specifications are subject to change without prior notice for improvement. 00YC2405