Data sheet
testo 890
FeverDetection kit

Thermal imager
testo 890 FeverDetection kit

testo FeverDetection function for identifying increased
surface temperatures on faces
Infrared resolution 640 x 480 pixels
Very good thermal sensitivity of < 40 mK (< 0.04 °C)
Optical and audible alarms°C

HDMI interface for transmission to an external monitor

With the thermal imager testo 890, body surface

The surface temperature of the face is automaticallytemperatures of individual persons can be measured quickly

recorded at the warmest spot in the face (usually theand reliably at heavily frequented facilities such as airports,

inner corner of the eye), and an alarm is triggered if it

railway stations or shopping centres.

exceeds a certain threshold value. This allows those with

The testo FeverDetection function in the thermal imager

an elevated body surface temperature to be identified

testo 890 can identify the relative body surface temperature

quickly and reliably, and to be isolated for a precise medicalof persons, or more accurately the temperature difference

examination.

between “healthy” people (with a normal body temperature)and “potentially ill” people (with increased bodytemperature).

www.testo.com

testo 890 FeverDetection kit

Order data / Technical data

1981 4064/msp/I/06.2020

testo 890 FeverDetection kit
testo 890 thermal imager with testo FeverDetection function in a robust case,including professional software (free download), SD card, USB cable, carrying strap,lens-cleaning cloth, mains unit, Li-ion rechargeable battery and headset

Order no. 0563 0890 X7

Imager equipment

Infrared resolution

640 x 480 pixels

Digital camera

Thermal sensitivity(NETD)< 40 mK at +30 °CLens

Field of view /min. focusing distance

42° x 32° / 0.1 m

Laser (laser classification
635 nm, Class 2)**

Geometric resolution(IFOV)

1.13 mrad

Image refresh rateFocus
Spectral range

33 Hz*
Automatic/manual
7.5 to 14 µm

Visual image output
Image size /min. focusing distance

3.1 MP / 0.5 m

Digital zoom

testo FeverDetection

4.3" LCD touchscreenwith 480 x 272 pixels
1 to 3 x

File format individual
image
Removable storage device

4Ambient conditions

Colour palettes

9 (iron, rainbow, rainbow HC,
cold-hot, blue-red, grey,
inverted grey, sepia, Testo)

Operating temperature
range

Emissivity/reflected
temperature settings
Transmission correction(atmosphere)

0.01 to 1 / manual

Storage temperature range
Air humidity*
Inside the EU, outside 9 Hz** Excepting USA, China and Japan*** Bluetooth only in the EU, Norway, Switzerland, USA, Canada, Colombia,Turkey, Japan, Russia, Ukraine, India, Australia-15 to +50 °C-30 to +60 °C
20 to 80% RH, non-condensing

Housing protection class(IEC 60529)

IP54

Vibration (IEC 60068-2-6)

2G

Physical features
WeightDimensions (L x W x H)

4

Fast-charging, Li-ion battery can bechanged on-site
4.5 hours

Mains operation±2 °C, ±2 % of reading(higher value applies)(±3°C of m.v. at -30 to -22°C)

SD card 2 GB
(approx. 1,500 to 2,000 images)

In instrument/in charging station(optional)

USB 2.0, Micro HDMI

Accuracy.bmt; export options in .bmp, .jpg, .png,.csv, .xls

Charging options

Video output-30 to +100 °C /0 to +350 °C (switchable)

4

Operating time

IR image / real image

Measuring range

Bluetooth*** / wired headset

Power supply

Display options

Measurement

Laser marker (not available when
FeverDetection is activated)

Image storage

Battery type

Image presentation
Image display

Voice recording

442° x 32°

Tripod mounting
Housing

1630 g253 x 132 x 111 mm
1/4" - 20 UNC
ABS

PC software
System requirements

Windows 10, Windows Vista,
Windows 7 (Service Pack 1), Windows 8,
USB 2.0 interface

Standards, tests
EU Directive

2004 / 108 / EC

www.testo.com

Subject to change, including technical modifications.

Infrared image output