HASOSWIR LIFT 160
Wavefront sensor
The Prodigy

Ultra-high resolution
SWIR range
Alignment-freecompatible

光貿易株式会社 〒113-0034 東京都文京区湯島３－１３－８湯島不二ビル３０１
メール: contact@hikari-trading.com ウェブサイト: http://www.hikari-trading.com
TEL: 03-3832-3117 FAX: 03-3832-3118
HASO SWIR LIFT 160 DATASHEET 2311

HASO SWIR LIFT 160
A great choice for themost demanding SWIR
optical metrology
applications, the HASOSWIR LIFT 160 wavefrontsensor provides thehighest resolution inSWIR.
This generation
features the newSpotTrackerTM technology.
It provides absolute
wavefront and tilt
information, eliminatingalignment requirements
for faster and easier
implementation.
Compatible with theOptical Engineer
Companion modularsystem: easily combine
the accessories youneed.

APPLICATIONS
Successfully used in the most demanding applications in opticalmetrology, microscopy, and laser diagnostics, the HASO SWIR LIFT
160 performs multiple functions :+ Optical manufacturing metrology+ Complex optics characterization+ Middle frequencies mirror surface characterization+ Optical quality control, metrology (LIDAR, free spacecommunication, Automotive, Space and defense)+ Predict the performance of optical systems in terms of focusing
capability or imaging quality+ Drive a wavefront corrector to correct for system aberrations+ Quantify the effects of temperature and gravity on system
performance

FEATURES+ Direct wavefront acquisition of highly converging and diverging
beams with an accuracy of λ/100 RMS, including astigmatism and highorder aberrations, and many other parameters, making it the perfect
instrument for any complex optics alignment+ Beam collimation with sensitivity > 1 km radius of curvature+ Control and adjustment of axial laser beam deviation > 3 μrad RMS ​+ Complex optics characterization in single or double pathconfiguration in combination with R-FLEX2 metrology systems orR-FLEX LA metrology platforms+ 3D MTF measurements

HASO SWIR LIFT 160 DATASHEET 2311

SPECIFICATIONS*OPERATING SPECS
Aperture dimension
Phase sampling
Maximum acquisition frequency
Calibrated wavelength range
Minimum power
External trigger
Operating system

9.3 x 7.4 mm²160 x 128150 Hz (USB 3.0) or 49 Hz (GigE)
1.05 - 1.70 µm
1 pW
TTL signal
Windows 10

OPTICAL SPECS
Repeatability
Absolute wavefront measurement accuracy
Spatial samplingTilt dynamic range
Focus dynamic range

λ/200 RMS
λ/100 RMS
58 µm
> ± 3°± 0.040 m to ± ∞MISC
Dimension
Weight for USB version
Working temperature
Interface
Power consumption

75 x 78 x 63 mm3 (USB 3.0)
250 g
15 - 30 °C
USB 3.0 or GigE
<5W

HASO
SWIR LIFT 160
Wavelength range (nm)Dynamic range ( λ PtV)
1000

100

HASO SWIR LIFT 160
Dynamic range at λ = 1550 nm

10Tetrafoil
22.5°

Tetrafoil
0°

5th Spherical

5th ComaY

5th ComaX

5th Astig.
45°

5th Astig.
0°

Trefoil30°

Trefoil
0°Spherical

Coma
YComa
X

Astig.
45°

Astig.
0°

Focus

Tilt
YTiltX

1*Subject to changes without further notice

DIMENSIONS** (mm)USB 3.0 model**

HASO SWIR LIFT 160 DATASHEET 2311

SOFTWAREWAVEVIEWTM
Metrology Software
WAVEVIEWTM is the most advanced
wavefront measurement andanalysis software.
It offers more than 150 features
and tools optimized for a wide
range of highly demanding
applications.
Options :+ Extensions for PSF, MTF andStrehl ratio+ Optional SDK in C/C++, LabVIEWand Python

WAVETUNETM
Adaptive Optics Software
WAVETUNETM is a unique softwarethat seamlessly combines
wavefront measurement andcorrection features with extensive
instrument diagnostics.
It is perfectly adapted to our HASO
wavefront sensors, ILAO STAR,MIRAO and mu-DM deformablemirrors, as well as to a wide rangeof active components.
Options :+ Optional SDK in C/C++, LabVIEW
and Python

HASO SWIR LIFT 160 DATASHEET 2311