HASO4126 VIS

Wavefront sensor
The Big Guy

Large pupil
Alignment-free
Versatile
compatible

光貿易株式会社 〒113-0034 東京都文京区湯島３－１３－８湯島不二ビル３０１
メール: contact@hikari-trading.com ウェブサイト: http://www.hikari-trading.com
TEL: 03-3832-3117 FAX: 03-3832-3118
HASO4 126 VIS DATASHEET 2311

HASO4 126 VISAPPLICATIONSThe HASO4 126 VIS
Shack-Hartmann
wavefront sensor
provides high-resolution
combined with a large
pupil for maximum
precision and versatility.
This generation features
the new SpotTrackerTM
technology. It provides
absolute wavefront and tilt
information, eliminatingalignment requirements
for faster and easier
implementation.
Compatible with theOptical Engineer
Companion modularsystem: easily combine
the accessories youneed.

Successfully used in the most demanding applications in opticalmetrology, microscopy, and laser diagnostics, the HASO4 126 VIS
performs multiple functions:+ Quantify the aberrations of optical systems+ Align the system to ensure that it performs at its best
+ Predict the optical system’s performance in terms of focusing
capability (PSF) or imaging quality (MTF)+ Quantify the effects of temperature and gravity on systemperformance+ Verify that the optics comply with specifications+ Directly measure the optical system’s wavelength dependency+ Pilot a wavefront corrector to change the system’s aberrations+ Check whether the optical mount overly distorts the optics

FEATURES+ Easy setup on any beam size thanks to the large 13.8 x 10.2 mm2pupil+ Direct wavefront acquisition of converging and diverging F/5 beams
with an accuracy of λ/100 RMS, including astigmatism and high-orderaberrations+ Beam collimation with an accuracy better than 300 m radius ofcurvature+ Control and adjustment of axial laser beam deviation better than
3 μrad RMS

HASO4 126 VIS DATASHEET 2311

SPECIFICATIONS*OPERATING SPECS
Aperture dimension
Number of microlenses
Maximum acquisition frequency
Calibrated wavelength range
Minimum power
External trigger
Operating system

13.8 x 10.2 mm²170 x 126
30 Hz (USB 3.0) or 8 Hz (GigE)
400 - 750 nm
0.15 nW
TTL signal
Windows 10

OPTICAL SPECS
Repeatability
Absolute wavefront measurement accuracy· λ between 400-600nm· λ between 600-750nm
Spatial sampling
Tilt dynamic range
Focus dynamic range≤ 6 nm RMS~ λ/100 RMS~ 80 µm> ± 3°± 0.010 m to ± ∞MISC
Dimensions (Height x Width x Length)
Weight for USB version
Working temperature
Interface
Power consumption

42 x 47 x 60 mm3 (USB 3.0)
185 g
15 - 30 °C
USB 3.0 or GigE
3.6 W

< λ/200 RMS

HASO4126 VISWavelength range (nm)Dynamic range ( λ PtV)
10000

1000

HASO4 126 VIS

Dynamic range at λ = 635 nm

100

10Tetrafoil
22.5°

Tetrafoil
0°

5th ComaY

5th ComaX

5th Astig.
45°

5th Astig.
0°

Trefoil30°

5th Spherical*Subject to changes without further notice

Trefoil0°

Spherical

Coma
Y

ComaX

Astig.
45°

Astig.
0°

Focus

Tilt
YTilt
X1

DIMENSIONS** (mm)USB 3.0 model**

HASO4 126 VIS DATASHEET 2311

SOFTWAREWAVEVIEWTM
Metrology Software
WAVEVIEWTM is the most advanced
wavefront measurement andanalysis software.
It offers more than 150 features
and tools optimized for a wide
range of highly demanding
applications.
Options :+ Extensions for PSF, MTF andStrehl ratio+ Optional SDK in C/C++, LabVIEW
and Python

WAVETUNETM
Adaptive Optics Software
WAVETUNETM is a unique softwarethat seamlessly combines
wavefront measurement andcorrection features with extensive
instrument diagnostics.
It is perfectly adapted to our HASO
wavefront sensors, ILAO STAR,MIRAO and mu-DM deformablemirrors, as well as to a wide rangeof active components.
Options :+ Optional SDK in C/C++, LabVIEW
and Python

HASO4 126 VIS DATASHEET 2311