HASO
SWIR

Wavefront sensor
The InGaAs

High accuracy
SWIR range
Alignment-freecompatible

光貿易株式会社 〒113-0034 東京都文京区湯島３－１３－８湯島不二ビル３０１
メール: contact@hikari-trading.com ウェブサイト: http://www.hikari-trading.com
TEL: 03-3832-3117 FAX: 03-3832-3118
HASO SWIR DATASHEET 2311

HASO SWIRAPPLICATIONS
A great choice forshort wavelength
infrared applications,
ideal for industry andlaboratories.
This generation features
the new SpotTrackerTM
technology. It provides
absolute wavefront and tilt
information, eliminatingalignment requirements
for faster and easier
implementation.
Compatible with theOptical Engineer
Companion modularsystem: easily combine
the accessories youneed.

Successfully used in the most demanding applications in opticalmetrology, microscopy, and laser diagnostics, the HASO SWIR
performs multiple functions:+ Quantify aberrations in optical systems for LIDAR, free-spacecommunication, space and defense, etc.
+ Align the system to ensure that it performs at its best
+ Predict the performance of optical systems in terms of focusing
capability (PSF) or imaging quality (MTF)+ Verify that the optics comply with specifications+ Directly measure the optical system’s wavelength dependency+ Drive a wavefront corrector to correct for system aberrations+ Check whether the optical mount overly distorts the optics+ Diagnostic of ultra-short-pulses with the Gated version

FEATURES+ Laser beam deviation control better than 3 µrad RMS+ Collimation diagnostic up to curvature radii over 300m
+ Live wavefront acquisition. Measurement accuracy λ/100 RMS
guaranteed for beams down to F/5​
+ True tilt measurement, curvature, astigmatism and high-orderaberrations quantification+ Optionnal Gated version with ultra-short exposure time feature tosynchronize with a pulsed laser.

HASO SWIR DATASHEET 2311

SPECIFICATIONS*OPERATING SPECS
Aperture dimension
Number of microlenses
Maximum acquisition frequency
Calibrated wavelength range
Minimum power
External trigger
Operating system

9.30 x 7.44 mm²40 x 32
150 Hz (USB 3.0) or 49 Hz (GigE)
0.9 - 1.7 µm
0.3 pW
TTL signal
Windows 10

OPTICAL SPECS
Repeatability
Absolute wavefront measurement accuracy
Spatial sampling
Tilt dynamic range
Focus dynamic range

λ/200 RMS
λ/100 RMS
~ 232.5 µm> ± 3°± 0.017 m to ± ∞MISC
Dimensions (Height x Width x Length)
Weight for USB version
Working temperature
Interface
Power consumption
Exposure time of Gated version

75 x 78 x 63 mm3250 g
15 - 30 °C
USB 3.0 or GigE
<5W
100 ns - 9 µs

HASO
SWIR
Wavelength range (nm)Dynamic range ( λ PtV)
1000

100

HASO SWIR

Dynamic range at λ = 1550 nm
10Tetrafoil
22.5°

Tetrafoil
0°

5th Spherical

5th ComaY

5th ComaX

5th Astig.
45°

5th Astig.
0°

Trefoil30°

Trefoil0°

Spherical

Coma
YComa
X

Astig.
45°

Astig.
0°FocusTilt
YTilt
X

1

*Subject to changes without further notice

DIMENSIONS** (mm)USB 3.0 model**

HASO SWIR DATASHEET 2311

SOFTWAREWAVEVIEWTM
Metrology Software
WAVEVIEWTM is the most advanced
wavefront measurement andanalysis software.
It offers more than 150 features
and tools optimized for a wide
range of highly demanding
applications.
Options :+ Extensions for PSF, MTF andStrehl ratio+ Optional SDK in C/C++, LabVIEW
and Python

WAVETUNETM
Adaptive Optics Software
WAVETUNETM is a unique softwarethat seamlessly combines
wavefront measurement andcorrection features with extensive
instrument diagnostics.
It is perfectly adapted to our HASO
wavefront sensors, ILAO STAR,MIRAO and mu-DM deformablemirrors, as well as to a wide rangeof active components.
Options :+ Optional SDK in C/C++, LabVIEW
and Python

HASO SWIR DATASHEET 2311