HASO4126 BROADBAND

Wavefront sensor
The Hulk

From UV to IR
Large pupil
Alignment-freecompatible

光貿易株式会社 〒113-0034 東京都文京区湯島３－１３－８湯島不二ビル３０１
メール: contact@hikari-trading.com ウェブサイト: http://www.hikari-trading.com
TEL: 03-3832-3117 FAX: 03-3832-3118
HASO4 126 BROADBAND DATASHEET 2311

HASO4 126 BROADBAND
APPLICATIONSThe HASO4 126
BROADBAND ShackHartmann wavefront
sensor provides highresolution combined witha large pupil and large
band for unmatchedprecision and versatility.
This generation features
the new SpotTrackerTM
technology. It provides
absolute wavefront and tilt
information, eliminatingalignment requirements
for faster and easier
implementation.
Compatible with theOptical Engineer
Companion modularsystem: easily combine
the accessories youneed.

Successfully used in the most demanding applications in opticalmetrology, microscopy, and laser diagnostics, the HASO4 126
BROADBAND performs multiple functions:+ Quantify the aberrations of optical systems+ Align the system to ensure that it performs at its best
+ Predict the optical system’s performance in terms of focusing
capability (PSF) or imaging quality (MTF)+ Quantify the effects of temperature and gravity on system
performance+ Verify that the optics comply with specifications+ Directly measure the optical system’s wavelength dependency+ Pilot a wavefront corrector to change the system’s aberrations+ Check whether the optical mount overly distorts the optics

FEATURES+ Easy wavefront measurement on the whole spectrum of thesensor: 350 - 1100 nm with n​o wavelength dependency+ Easy setup on any beam size thanks to the large 13.8 x 10.2 mm2pupil+ Direct wavefront acquisition of converging and diverging F/5 beams
with an accuracy of λ/100 RMS, including astigmatism and high-orderaberrations+ Beam collimation with an accuracy better than 300 m radius ofcurvature+ Control and adjustment of axial laser beam deviation better than
3 μrad RMS

HASO4 126 BROADBAND DATASHEET 2311

SPECIFICATIONS*OPERATING SPECS
Aperture dimension
Number of microlenses
Maximum acquisition frequency
Calibrated wavelength range
Minimum power
External trigger
Operating system

13.8 x 10.2 mm²170 x 126
30 Hz (USB 3.0) or 8 Hz (GigE)
350 - 1100 nm
0.15 nW
TTL signal
Windows 10OPTICAL SPECS
Repeatability
Absolute wavefront measurement accuracy· λ between 350-600 nm· λ between 600-1100 nm
Spatial sampling
Tilt dynamic range
Focus dynamic range≤ 6 nm RMS~ λ/100 RMS~ 80 µm> ± 3°± 0.010 m to ± ∞MISC
Dimensions (Height x Width x Length)
Weight for USB version
Working temperatureInterface
Power consumption

42 x 47 x 60 mm3 (USB 3.0)
185 g
15 - 30 °C
USB 3.0 or GigE
3.6 W

< λ/200 RMS

HASO4
126 BROADBAND
Wavelength range (nm)Dynamic range ( λ PtV)
10000

1000

HASO4 126 BROADBAND
Dynamic range at λ = 635 nm

100

10Tetrafoil
22.5°

Tetrafoil
0°

5th ComaY

5th ComaX

5th Astig.
45°

5th Astig.
0°

5th Spherical*Subject to changes without further notice

Trefoil30°

Trefoil0°

Spherical

Coma
Y

Coma
X

Astig.
45°

Astig.
0°

Focus

Tilt
YTilt
X

1

DIMENSIONS** (mm)USB 3.0 model**

HASO4 126 BROADBAND DATASHEET 2311

SOFTWAREWAVEVIEWTM
Metrology Software
WAVEVIEWTM is the most advanced
wavefront measurement andanalysis software.
It offers more than 150 features
and tools optimized for a wide
range of highly demanding
applications.
Options :+ Extensions for PSF, MTF andStrehl ratio+ Optional SDK in C/C++, LabVIEW
and Python

WAVETUNETM
Adaptive Optics Software
WAVETUNETM is a unique softwarethat seamlessly combines
wavefront measurement andcorrection features with extensive
instrument diagnostics.
It is perfectly adapted to our HASO
wavefront sensors, ILAO STAR,MIRAO and mu-DM deformablemirrors, as well as to a wide rangeof active components.
Options :+ Optional SDK in C/C++, LabVIEW
and Python

HASO4 126 BROADBAND DATASHEET 2311