HASO

LIFT 680
Wavefront sensor
The Best-in-class

Ultra-high spatial resolution
High accuracy
Alignment-freecompatible

光貿易株式会社 〒113-0034 東京都文京区湯島３－１３－８湯島不二ビル３０１
メール: contact@hikari-trading.com ウェブサイト: http://www.hikari-trading.com
TEL: 03-3832-3117 FAX: 03-3832-3118
HASO LIFT 680 DATASHEET 2311

HASO LIFT 680
For the HASO LIFT 680,
Imagine Optic has
merged the reliability
and accuracy of a ShackHartmann wavefront
sensor with the ultrahigh resolution of LIFT.
This generation features
the new SpotTrackerTM
technology. It provides
absolute wavefront and tilt
information, eliminatingalignment requirements
for faster and easier
implementation.
Compatible with theOptical Engineer
Companion modularsystem: easily combine
the accessories youneed.

APPLICATIONS
Successfully used in the most demanding applications in opticalmetrology, microscopy, and laser diagnostics, the HASO LIFT 680
performs multiple functions :+ Characterize complex optics, including meta-surface and freeform
optics+ Quantify laser impact (LIDT)+ Perform surface characterization on high and middle frequencies
mirrors+ Predict the performance of optical systems in terms of focusing
capability or imaging quality+ Quantify the effects of temperature and gravity on system
performance+ Drive a wavefront corrector to correct for system aberrations

FEATURES
HASO LIFT 680 enables you to perform multiple functions by
combining :​+ Ultra-high spatial resolution of 680 x 504, allowing characterization
over several hundreds of Zernike polynomials+ Accuracy of λ/100 RMS permitting small defects detection+ Dynamic range superior to 1000 λ for direct wavefront acquisitionof converging and diverging beams

HASO LIFT 680 DATASHEET 2311

SPECIFICATIONS*OPERATING SPECS
Aperture dimension
Phase points resolution
Number of microlenses
Maximum acquisition frequency
Calibrated wavelength range
Minimum power
External trigger
Operating system

13.8 x 10.2 mm²680 x 504
170 x 126
30 Hz (USB 3.0) or 8 Hz (GigE)
400 - 750 nm
0.7 nW
TTL signal
Windows 10

OPTICAL SPECS
Repeatability
Absolute wavefront measurement accuracy
Spatial sampling
Tilt dynamic range
Focus dynamic range< λ/200 RMS
λ/100 or 6 nm RMS~ 20 µm> ± 3°± 0.004 m to ± ∞MISC
Dimension (Height x Width x Length)
Weight for USB version
Working temperature
Interface
Power consumption

47 x 62 x 60 mm3 (USB 3.0)
200 g
15 - 30 °C
USB 3.0 or GigE
3.6 W

HASO
LIFT 680
Wavelength range (nm)Dynamic range ( λ PtV)
10000

1000

HASO LIFT 680

Dynamic range at λ = 635 nm

100

10Tetrafoil
22.5°

Tetrafoil
0°

5th ComaY

5th ComaX

5th Astig.
45°

5th Astig.
0°

5th Spherical*Subject to changes without further notice

Trefoil30°

Trefoil0°

Spherical

Coma
YComa
X

Astig.
45°

Astig.
0°

Focus

Tilt
YTilt
X

1

DIMENSIONS (mm)**USB 3.0 model**

HASO LIFT 680 DATASHEET 2311

SOFTWAREWAVEVIEWTM
Metrology Software
WAVEVIEWTM is the most advanced
wavefront measurement andanalysis software.
It offers more than 150 features
and tools optimized for a wide
range of highly demanding
applications.
Options :+ Extensions for PSF, MTF andStrehl ratio+ Optional SDK in C/C++, LabVIEW
and Python

WAVETUNETM
Adaptive Optics Software
WAVETUNETM is a unique softwarethat seamlessly combines
wavefront measurement andcorrection features with extensive
instrument diagnostics.
It is perfectly adapted to our HASO
wavefront sensors, ILAO STAR,MIRAO and mu-DM deformablemirrors, as well as to a wide rangeof active components.
Options :+ Optional SDK in C/C++, LabVIEW
and Python

HASO LIFT 680 DATASHEET 2311