Pattern Generator Module
PN L-6001-EPG10-6
DESCRIPTION
EPG10-6 is an Electrical Pattern Generator module that plugs into theXBERT and ParalleX™ Chassis. EPG10-6 can generate electrical datafrom 1 Gb/s up to 13 Gb/s. User programmable PRBS patterns can bechanged via an easy to use GUI. A pattern trigger output provides anelectrical trigger synchronous with the pattern for use with an oscilloscopeor other test equipment. Front panel indicators give immediate status forTx Data.

KEY FEATURES
Data Rates 1 to 13 Gbps

Differential Electrical Pattern Generator (SMA Connector)



Output voltage adjustable from 0.3V to 1.0V in 0.1V steps



PRBS:
7, 9, 10, 11, 15, 21, 23, 31
User-Pattern: 8Bit - 8Kbyte
1111
Clock-Pattern: /1, /2, /4, /8
Additional:
K28.0-K28.7, CJPAT, SSPS-64 and others

Data input & output polarity swap

Single error and error rate injection: 1E-3 to 1E-15



Data log Gating-time: up to 5000h

High speed Clock Input and Output

Trig in and Trig outLabView

GPIB/LAN/USB Interface via XBERT Chassis.

Small size: width 25.4mm (1”)

TM

PATTERN GENERATOR MODULE PN L-6001-EPG10-6

FUNCTIONAL BLOCK DIAGRAM

ClockInput

High speed
Clock Input
Single Ended

PatternGenerator

Data Output
Differential orSingle Ended

EPG10-6

High speed
Clock Output
Single Ended

ModuleTrig Output

Productdrivers available
Clock
DomainTrig Input

CTRL via
GPIB, LAN, USB

XBERT PLATFORM: LETS YOU START SMALL AND GROWXBERT is a low-cost, modular Bit Error Rate Test
Platform used for verification and test of up to 13 Gbps
optical and electrical chip, sub assembly and systemdesigns. ParalleX® allows users to perform several BER
tests at once using a single clock source. The system isideal for developers desiring to run simultaneous BER
tests on parallel interfaces or multiple independentinterfaces. XBERT and ParalleX® are scalable so users
can start off with a single channel and add modules togrow the system. Manufacturers can realize great savings
by taking advantage of the XBERT and ParalleX®
system’s scalability to perform parallel testing in volume
production environments.

PRELIMINARYPB EPG10-6 Rev. A.01 Issued date 2010.05.27All specifications are subject to change without notice

Overview

Pattern Generator Module PN L-6001-EPG10-6
KEY PERFORMANCE PARAMETERSPARAMETERSYMBOLMinMax

UNIT

Data Rate

DR113Gbps

Data Formats

NRZ
7, 9, 10, 11, 15, 21,23, 31

PRBS PatternUser-Defined Pattern lengthSingle ended Data Output Amplitude

NOTE

DOutP/N

865536Bit

300

1000mVpp

100mV

Output Voltage Step Sizeadjustable

Dout on/off

D on/off

4001100mVpp

SMA input toswitch on/off theDout signal

Data Output Rise and Fall timetr / tf1220

ps

20% - 80%

Output Jitter (RMS)

Jrms

1ps

Output Jitter (DJ)

Jdj

5ps

Single ended Data Output
Impedance

ZOse55

45

Data Output Termination

AC - coupled

Clock Input Amplitude3001000mVpp

Clock Output Amplitude3001000mVpp

13GHz55



Clock Input / Output Frequency

FClk

1

Clock Input / Output Impedance

ZClk

45

Clock Input / Output Termination

Not adjustable

AC - coupled

Pattern Output Trigger

Pout

550

1100

mV

Sync signal for
pattern start

Pattern Input Trigger

Pin4001100mV

Signal to start
the pattern

Operating Temperature

TOP

040°C

Ambient temp.

EPG10-6
ModuleProductOverview

ELECTRICAL EYE-DIAGRAM
PRBS31, Data Rate = 12.5Gbps, output voltage = 1.0V

Am Borsigturm 1313507 Berlin-Germany+49 30 814 512 40 00 ph+49 30 814 512 40 20 fax
www.luceotec.com
sales@luceotec.com