Relative and differential
pressure transmitter type 692
Pressure range
0 ... 0.1 25 bar

Type 692 pressure transmitters have a unique, well provenceramic technolgy. There are variety of pressure and electricalconnections available, together with several stanadardised output
signals. The wide variety of options makes these transmittersideal for applications across a broad spectrum of industries.

Huba Control type 692 - Technical data subject to change - Edition 04/2017

Very low temperature sensitivity
High resistance to extreme temperatures
No mechanical creepage
Modular system and choice of materialsto suit individual applications

1/5Technical overview
Pressure range
Relative and differential

0 ... 0.1

Operating conditions
MediumMedium / ambient
Storage

Temperature
Tolerable overload on one sideSystem pressure< 6 bar

PVDF

Stainless steel 1.4305 / AISI 303> 10 barRupture pressure
Materials
Case
Pressure connection
Materials in contact with the medium

SensorSealing material

25 barLiquids and neutral gases-15 ... +85 ÀC-40 ... +85 ÀC
See order code selection table
12 bar
25 bar
50 bar
1.5x system pressure

Stainless steel 1.4305 / AISI 303
Stainless steel 1.4305 / AISI 303,
PVDF, CuZn nickel plated
ceramic Al2O3 (96%)
FPM, EPDM, NBR, MVQ

Electrical overview
2 wire3 wirePolarity reversal protection
Dynamic response
Response time
Load cycle

OutputPower supply
Load
Current consumption (at nominal pressure)- 11 V
4 ... 20 mA
11 ... 33 VDC< supply voltage[Ohm]< 20 mA
0.02 A
0 ... 5 V
11 ... 33 VDC / 24 VAC μ15%
>10 kOhm< 5 mA
0 ... 10 V
18 ... 33 VDC / 24 VAC μ15%
>10 kOhm< 5 mAratiom. 10 ... 90%
5 VDC μ5%>10 kOhm
< 5 mA
Short circuit proof and protected against polarity reversal. Each connection is protected against crossover up to max. supply voltage.

< 5 ms< 50 Hz

Protection standard
IP 65
Electrical connection
Connector DIN EN 175301-803-A
Connector DIN EN 60130-9
Cable 1.5 m

PG7Pressure connection
Pressure tube tip
Pipe fitting
Outside thread
Inside thread± 4 mm± 6 mm± 6 mm± 8 mm7/16 -20 UNF
G⅛⅛ -27 NPT
G⅛

Mounting instruction
Installation arrangement
Mounting

Unrestricted
Mounting bracket

Tests / Admissions
Electromagnetic compatibility

CE conformity acc. EN 61326-2-3

Weight~ 430 gPackaging
Single packaging in cardboard

accessories included

Legend to cross-section drawing
1 Electrical connection
2 Seals
3 Ceramic element
4 P2 Pressure connection (lower pressure)
5 P1 Pressure connection (higher pressure)

2/5

Huba Control type 692 - Technical data subject to change - Edition 04/2017

Accuracy
Parameter

Unit

Tolerance zero pointmax.
Tolerance full scalemax.
Resolution
Total of linearity,hysteresis and repeatability
max.
Long term stability acc. to DIN EN 60770
TC zero point 1)
max.
TC sensitivity 1)
max.
Test conditions:% fs% fs% fs% fs% fs% fs/10K% fs/10K

Versions with overload on one
side < 2x nominal pressureμμVersions with overload on one
side < 3x nominal pressure

0.4
0.4
0.1

μμ

μ0.5
μ0.5
See order code selection table
μ0.15

Versions with overload on one side< 7.5x nominal pressure

0.75
0.75
0.15

μμ

μ0.75
μ0.5
See order code selection table
μ0.23

1.25
1.25
0.25

μ1.25
μ0.5
See order code selection table
μ0.3825 ÀC, 45% RH, Power supply 24 VDC
TC z.p. / TC s. -15 ... +80 ÀC
1Order code selection table

Pressure range 2)

Sealing material
Adjustment
Output / power supply

Electrical connection

Pressure connection

Case
Pressure range variation(optional)

2345678910692. X X X X X X X X X X

Tolerable overload on one side
P1
0 ... 0.1 bar
max. 0.6 bar(6 x Nominal pressure)
0 ... 0.2 bar
max. 1.2 bar(6 x Nominal pressure)
0 ... 0.2 bar
max. 0.6 bar(3 x Nominal pressure)
0 ... 0.25 bar
max. 1.2 bar(4.8 x Nominal pressure)
0 ... 0.25 bar
max. 0.6 bar(2.4 x Nominal pressure)
0 ... 0.3 bar
max. 0.6 bar(2 x Nominal pressure)
0 ... 0.4 bar
max. 1.2 bar(3 x Nominal pressure)
0 ... 0.4 bar
max. 2 bar(5 x Nominal pressure)
0 ... 0.5 bar
max. 1.2 bar(2.4 x Nominal pressure)
0 ... 0.5 bar
max. 3 bar(6 x Nominal pressure)
0 ... 0.6 bar
max. 1.2 bar(2 x Nominal pressure)
0 ... 0.6 bar
max. 3 bar(5 x Nominal pressure)
0 ...
1 bar
max. 2 bar(2 x Nominal pressure)
0 ...
1 bar
max. 5 bar(5 x Nominal pressure)
0 ... 1.6 bar
max. 3.2 bar(2 x Nominal pressure)
0 ... 1.6 bar
max. 12 bar(7.5 x Nominal pressure)
0 ... 2.5 bar
max. 5 bar(2 x Nominal pressure)
0 ... 2.5 bar
max. 12 bar(4.8 x Nominal pressure)
0 ...
4 bar
max. 8 bar(2 x Nominal pressure)
0 ...
4 bar
max. 12 bar(3 x Nominal pressure)
0 ...
6 bar
max. 12 bar(2 x Nominal pressure)
0 ... 10 bar
max. 20 bar(2 x Nominal pressure)
0 ... 16 barmax. 32 bar(2 x Nominal pressure)
0 ... 25 bar
max. 50 bar(2 x Nominal pressure)▲ Fullscale signal at these pressures
FPM
Fluoro elastomer
EPDM
Ethylene propylene
NBR
Butadiene Acrylonitrile
MVQ
Silicone polymer
Factory
0 ... 5 V
11 ... 33 VDC / 24 VAC μ15%
0 ... 10 V
18 ... 33 VDC / 24 VAC μ15%
4 ... 20 mA
11 ... 33 VDC
ratiom. 10 ... 90% 5 VDC μ5%
Cable 1.5 m, PG7
DIN EN 175301-803-A
Connector 3)
DIN EN 60130-9
Inside thread
Stainless steel ⅛ -27 NPT or PVDF G ⅛CuZn nickel plated
Stainless steel 1.4571 / AISI 316Ti
Hose connection
CuZn nickel plated
PVDF
Stainless steel 1.4571 / AISI 316Ti
CuZn nickel plated
Stainless steel 1.4305 / AISI 303
PVDF
Pipe fitting
CuZn nickel plated
Stainless steel 1.4305 / AISI 303
PVDF
7Outside thread/16 -20 UNF CuZn nickel plated
Adapter inside
G ⅛ Stainless steel 1.4305 / AISI 303
Adapter outside
G ⅛ CuZn nickel plated with union nut
Stainless steel1.4305 / AISI 303
PVDF to 6 bar max.
Stainless steel with pressure tip orifice

TC z.p. (fs/10K)
P20.6 bar
1.2 bar
0.6 bar
1.2 bar
0.6 bar
0.6 bar
1.2 bar2 bar
1.2 bar
3 bar
1.2 bar
3 bar
2 bar
5 bar
3.2 bar
12 bar
5 bar
12 bar8 bar
12 bar
12 bar
20 bar
32 bar
50 barμ 1.2 %
μ 1.2 %
μ 0.6 %
μ 1.0 %
μ 0.5 %
μ 0.4 %
μ 0.6 %
μ 1.0 %
μ 0.5 %
μ 0.8 %
μ 0.4 %
μ 0.7 %
μ 0.4 %
μ 1.0 %
μ 0.4 %
μ 1.0 %
μ 0.4 %
μ 0.6 %
μ 0.4 %
μ 0.5 %
μ 0.4 %
μ 0.4 %
μ 0.4 %
μ 0.4 %

9999999999999999999999990040400000001111111113330203114567891234567890121,41,41,401230017901301E23D458679AB
C

for tube inside ± 4 mm
for tube inside ± 4 mm
for tube inside ± 6 mm
for tube inside ± 6 mm
for tube inside ± 6 mm
for pipe outside ± 6 mm
for pipe outside ± 6 mm
for pipe outside ± 6 mm
for pipe outside ± 8 mm
for pipe outside ± 8 mm
for pipe outside ± 8 mm

Indicate W and state range on order (e.g.: W0... + 8bar/OUT1...6V)

1,4
1,4
1,4
2
1,4
1,4
1,4
2
1,4
1,4
2
1,4
1,41,4124W

Accessories
Order number
Female connector DIN EN 175301-803-A with seal
Female connector DIN EN 60130-9
Mounting bracket incl. screws
Calibration certificate1)

TC = Temperature coefficientIP 65, when installed and screwedIP 65, when installed and screwed2)

Other pressure range on request

Huba Control type 692 - Technical data subject to change - Edition 04/2017

1035101035241019991045513)

Delivery without female connector

3/5Dimensions in mm / Electrical connections
Female Connector
DIN EN 60130-9

Female connector DIN EN
175301-803-A
Cable

Stainless steel

Stainless steel ~138 / PVDF ~132

Stainless steel ~95 / PVDF ~89

Stainless steel ~158 / PVDF ~152 (mounted)

Stainless steel ~108 / PVDF ~102

Stainless steel ~136 / PVDF ~130 (mounted)

PVDF

BracketScrew fitting for pipe outside ± 6

L~24X~65

a=10b=12

Screw fitting for pipe outside ± 8

L~26
X~67

a=12b=14L~12X~53

a=14

Screw fitting for pipe outside ± 6

L~24X~65

a=10b=12

Screw fitting for pipe outside ± 8

L~25
X~66

a=12b=14

CuZnnickel plated Hose connection for tube ± 4
Stainless steel
1.4571
Hose connection for tube ± 6
AISI 316Ti

L~20
X~61
L~25
X~66

a=10

CuZn
Outside thread G ⅛
nickel plated

L~20X~61

a=10b=12

Stainlesssteel
1.4305AISI 303

Stainless Steel1.4305Inside thread G ⅛
AISI 303

CuZn
nickel plated

a=102 wire
brown
Cable

L~18
X~59

a=14

Screw fitting for pipe ± 6

L~20
X~61

a=12

Screw fitting for pipe ± 8

L~23
X~64

a=14

Hose connection for tube ± 6

L~20
X~61

a=10

PVDF

PVDF

4/5

brown
whitegreen
CuZn
Outside thread 7/16 -20 UNF
nickel plated

3 wiregreen

Connector DIN
EN 175301-803-A

Connector DIN
EN 60130-9

Huba Control type 692 - Technical data subject to change - Edition 04/2017

5/5

Huba Control type 692 - Technical data subject to change - Edition 04/2017