Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

The image shows model OE-300-IN-01-FC.
28Features••
••••
•
•

Adjustable transimpedance gain from 10 to 10 V/A
Wide bandwidth up to 200 MHz
InGaAs-PIN photodiode covering the 900 to 1700 nm wavelength range
FC fiber optic input
High dynamic input range up to 10 mW optical power
Very low noise, NEP down to 47 fW/√HzSwitchable low pass filters for minimizing wideband noise
Full manual and remote control capabilityApplications••••

All-purpose low-noise photoreceiver (O/E converter) for the MHz range
Time resolved optical pulse and power measurements
Laser intensity noise measurements (RIN)
Optical front-end for oscilloscopes, spectrum analyzers, A/D converters andRF lock-in amplifiers
Current to voltage converter
Rf = 100 Ω ... 10 MΩ

Block Diagram

Buffer-amplifier andbandwidth limiting

Programmable
AC / DC coupling

OPTICAL
INPUT

Programmablegain amplifier

Offsetnulling

X 10I/V

VOLTAGEOUTPUT

X1

10 MHzFBW
1 MHzHigh speed
Low noise
Stabilizedbias voltage

Overloaddetector
DCMONITOROUTPUT

Supplyvoltageregulator

Manualswitches
Parameter
control unit
Optocoupler
Isolated unit

DIGITALCONTROL
INPUTSPOWER
SUPPLYINPUT

BS-OE-300-R1

SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
DE-OE-300-IN-01_R1/MG,JM/20JAN2016

Page 1 of 11

Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Available Versions

Related OE-300 Models

Available Accessories

OE-300-IN-01-FC

FC fiber optic input

See separate datasheets for following models on www.femto.de:OE-300-SI-10-FST

Si-PIN, 1 mm x 1 mm, 400 - 1000 nm
1.035”-40 threaded flange

OE-300-SI-10-FS

Si-PIN, 1 mm x 1 mm, 400 - 1000 nm
25 mm dia. unthreaded flange

OE-300-SI-30-FSTSi-PIN, ø 3 mm, 320 - 1000 nm
1.035”-40 threaded flange

OE-300-SI-30-FS

Si-PIN, ø 3 mm, 320 - 1000 nm
25 mm dia. unthreaded flange

OE-300-IN-03-FST

InGaAs-PIN, ø 300 µm, 800 - 1700 nm
1.035”-40 threaded flange

OE-300-IN-03-FS

InGaAs-PIN, ø 300 µm, 800 - 1700 nm
25 mm dia. unthreaded flange

OE-300-S

customized versions available on request

PRA-PAP

post adapter plate,easy to mount on
FEMTO photoreceiver seriesOE, FWPR, HCA-S and LCA-S

PS-15

power supply,input: 100 - 240 VAC,
output: ±15 VDC, +400/–250 mA

LUCI-10compact digital I/O interface for USB remote control,
supports opto-isolation of amplifier signal path from PC
USB port, 16 digital outputs, 3 opto-isolated digital inputs,
bus-powered operation

SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 2

Datasheet

OE-300-IN-01
200 MHz Variable Gain PhotoreceiverSpecifications

Test conditions

VS = ±15 V, TA = 25 °C, system impedance = 50 ΩGain

Transimpedance gain
Gain accuracy

1 x 10 ... 1 x 10 V/A±1 %Frequency Response

Lower cut-off frequency
Upper cut-off frequency

DC/100 Hz, switchable
up to 200 MHz (see table below),switchable to 1 MHz or 10 MHz

Input

Noise equivalent power (NEP)
Max. CW saturation power

see table below
see table below

Detector

Detector
Active area

InGaAs-PIN photodiodeIntegrated ball lens, suitable for fibers up to
62.5 µm core diameter

Spectral response
Sensitivity R
Dark current

900 - 1700 nm
0.95 A/W typ. @ 1550 nm
0.02 nA typ.

Gain setting (low noise) (V/A)

10Performance Depending
on Gain Setting

228103104105106107

Upper cut-off frequency (–3 dB)
NEP (/√Hz, @ 1550 nm)
Measured at
Integrated input noise (RMS)*
CW sat. power (@ 1550 nm)

200 MHz 80 MHz
180 pW 22 pW
20 MHz 8 MHz
4.9 µW 380 nW
10 mW 1.0 mW

Gain setting (high speed) (V/A)

10310414 MHz
1.9 pW
1.4 MHz23 nW
100 µW
1053.5 MHz
390 fW
350 kHz3.3 nW
10 µW
1061.8 MHz 220 kHz
140 fW 50 fW
180 kHz 22 kHz
0.84 nW 71 pW1.0 µW 100 nW
107108

Upper cut-off frequency (–3 dB)
NEP (/√Hz, @ 1550 nm)
Measured at
Integrated input noise (RMS)*
CW sat. power (@ 1550 nm)

175 MHz 80 MHz
132 pW 6.3 pW
18 MHz 8 MHz
3.0 µW 285 nW
1.0 mW 100 µW

14 MHz1.4 pW
1.4 MHz
21 nW
10 µW

3.5 MHz350 fW
350 kHz3.2 nW
1.0 µW

1.8 MHz 220 kHz
113 fW 47 fW
180 kHz 22 kHz
0.84 nW 71 pW
100 nW 10 nW* The integrated input noise is measured with a shaded input in the full bandwidth (‘‘FBW’’) setting(referred to 1550 nm). The measurement bandwidth is 3 x the upper cut-off frequency at thest
specific gain setting; filter slope is a 1 order roll-off.
The input referred peak-peak noise can be calculated from the RMS noise as follows:
PInput noise peak-to-peak= PInput noise RMS x 6
The output noise is given by:

UOutput noise RMS
UOutput noise peak-to-peak= PInput noise RMS x gain x R= UOutput noise RMS x 6 = PInput noise RMS x gain x R x 6

The integrated noise will be reduced considerably by setting the low pass filter to ‘‘1 MHz’’ or‘‘10 MHz’’ instead of ‘‘FBW’’. This is especially useful for continuous wave (CW) measurements.

SOPHISTICATED TOOLS FOR SIGNAL RECOVERYPage 3

Datasheet

OE-300-IN-01
200 MHz Variable Gain PhotoreceiverSpecifications (continued)
OutputOutput voltage range
Output impedance
Slew rate
Max. output current
Output offset compensation±1 V (@ 50 Ω load), for linear amplification
50 Ω (designed for 50 Ω load)
1000 V/µs±40 mAadjustable by offset potentiometer and external controlvoltage, output offset compensation range min. ±100 mV

Ext. Offset Control

Control voltage range
Offset control input impedance±10 V
15 kΩIndicator LEDFunctionoverload

Digital Control

Control input voltage range
Control input current
Overload output

LOW bit: –0.8 ... +1.2 V, HIGH bit: +2.3 ... +12 V
0 mA @ 0 V, 1.5 mA @ +5 V, 4.5 mA @ +12 V
non active: <0.4 V @ 0 ... –1 mA
active: typ. 5 ... 5.1 V @ 0 ... 2 mA

Power Supply

Supply voltage
Supply current
Stabilized power supply output±15 V+110/–90 mA (depends on operating conditions,recommended power supply capability min ±200 mA)±12 V, max. 20 mA, +5 V, max. 150 mA

CaseWeightMaterial

320 g (0.74 lb.)
AlMg4.5Mn, nickel-plated

DC Monitor Output

Monitor output gain

Mode

Monitor gainLow noiseHigh speed

Gain setting divided by –1
Gain setting divided by –10

Monitor output polarity
Monitor output voltage range
Monitor output bandwidth
Monitor output impedance

inverting±1 V (@ ≥1 MΩ load)
DC ... 1 kHz
1 kΩ (designed for ≥1 MΩ load)

Temperature Range

Storage temperature
Operating temperature–40 ... +80 °C
0 ... +60 °C

Absolute MaximumRatings

Max. CW power (averaged)
Digital control input voltage
Analog control input voltage
Power supply voltage

12 mW–5 V/+16 V relative to digital ground DGND (pin 9)±15 V relative to analog ground AGND (pin 3)±20 V

SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 4

Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Connectors

Input

FC fiber optic receptacle

Output

BNC jack (female)Power supply

Lemo series 1S, 3-pin fixed socket(mating plug type: FFA.1S.303.CLAC52)
Pin 1:+15 V
Pin 2:–15 V
Pin 3:GND®

PIN 2–Vs

PIN 1+Vs

PIN 3
GND

Control port

Scope of Delivery

Sub-D 25-pin, female, qual. class 2
Pin 1:+12 V (stabilized power supply output)
Pin 2:–12 V (stabilized power supply output)
Pin 3:
AGND (analog ground for pins 1 - 8)
Pin 4:+5 V (stabilized power supply output)
Pin 5:
digital output: overload (referred to pin 3)
Pin 6:DC Monitor output
Pin 7:NC (= not connected)
Pin 8:
output offset control voltage input
Pin 9:
DGND (ground for digital control pins 10 - 16)
Pin 10:digital control input: gain, LSB
Pin 11:
digital control input: gain
Pin 12:
digital control input: gain, MSB
Pin 13:
digital control input: AC/DC
Pin 14:
digital control input: high speed / low noise
Pin 15:upper cut-off frequency limit 10 MHz
Pin 16:
upper cut-off frequency limit 1 MHz
Pin 17 - 25: NC (= not connected)®

OE-300-IN-01-FC, Lemo 3-pin connector, datasheet, transport package

SOPHISTICATED TOOLS FOR SIGNAL RECOVERYPage 5

Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Remote Control Operation

GeneralRemote control input bits are opto-isolated and connectedby a logical OR function to the local switch settings. For
remote control set the corresponding local switches to“Remote”, “AC” and “H” and select the desired setting via
a bit code at the corresponding digital inputs.
Mixed operation, e.g. local AC/DC setting and remotecontrolled gain setting, is also possible.

Gain setting

Low noiseHigh speed
Gain (V/A)
Gain (V/A)
Pin 12
Pin 14=HIGH Pin 14=LOW MSB

Pin 11

Pin 10LSB
2AC/DC setting

31031041051061071010410510610710810CouplingPin 13

LOW
LOW
LOW
LOW
HIGH
HIGH

LOW
LOW
HIGH
HIGH
LOW
LOW

Pin 15Pin 16

LOW
HIGH
LOW
HIGH
LOW
HIGH

DCAC
Low pass filter setting

LOW
HIGH

Upper cut-off freq. limitHigh speed / low noise setting

full bandwidth
10 MHz
1 MHz

LOW
HIGH
LOWModePin 14

LOW
LOW
HIGH

low noise mode
LOW

high speed modeHIGH

Spectral Responsivity

SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 6

Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Typical Performance
CharacteristicFrequency response
VSupply = ±15 VDC; RLoad = 50 Ω23Gain setting: L10 , H1045Gain setting: L10 , H1067

Gain setting: L10 , H10Gain setting: L10 , H10Gain setting: L10 , H10Gain setting: L10 , H10345678SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 7

Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Typical PerformanceCharacteristic (continued)

Input noise equivalent power (NEP)
23Gain setting L10 , H1045Gain setting: L10 , H1067Gain setting: L10 , H10Gain setting L10 , H10Gain setting: L10 , H10Gain setting: L10 , H10345678SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 8

Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Typical Performance
Characteristic (continued)

Signal pulse response
Gain setting L102Gain setting H103Gain setting L103Gain setting H104Gain setting L104Gain setting H105SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 9Datasheet

OE-300-IN-01
200 MHz Variable Gain Photoreceiver

Typical Performance
Characteristic (continued)

Gain setting L105Gain setting H106Gain setting L106Gain setting H107Gain setting L107Gain setting H108SOPHISTICATED TOOLS FOR SIGNAL RECOVERY
Page 10

Datasheet

OE-300-IN-01
200 MHz Variable Gain PhotoreceiverDimensions

Fiber optic input OE-300-IN-01-FC:

157 mm
150 mm

15 mm

137 mm

GAIN

INOUT

HIGH SPEED
OFFSET

10 MHzACPOWERFBWLOW NOISE

1 MHz

51 mm

44 mm

OVERLOADDC43.5 mm

28.5 mm

7.8 mmØ 3.2 mm

DZ-OE-300-FC_R2

FEMTO Messtechnik GmbHKlosterstr. 6410179 Berlin · Germany
Phone: +49 30 280 4711-0
Fax: +49 30 280 4711-11
Email: info@femto.de
www.femto.deSpecifications are subject to change without notice. Information provided herein is believed to be accurate andreliable. However, no responsibility is assumed by FEMTO Messtechnik GmbH for its use, nor for any infringementof patents or other rights of third parties which may result from its use. No license is granted by implication orotherwise under any patent or patent rights of FEMTO Messtechnik GmbH. Product names mentioned may also betrademarks used here for identification purposes only.
© by FEMTO Messtechnik GmbH · Printed in Germany

SOPHISTICATED TOOLS FOR SIGNAL RECOVERYPage 11