PRODUCT DATA

GSS 5GV Vibration Monitor
FOR STRUCTURAL AND OCCUPANT MONITORING
The 5GV is a cutting-edge MEMS Vibration andTilt monitor that boasts exceptional precision andan integrated 3G/4G gateway, making it the ideal
solution for a wide range of industries. Its userfriendly design ensures that it is easy to configure,install, and manage remotely, reducing the needfor time-consuming on-site visits. Thanks to its
low-power design and external power options, the5GV minimizes maintenance activities, allowing youto focus on what really matters. With user-definedalert thresholds, you will receive notifications directly
when something requires your attention.

Features— Standalone Sensor with Internal Storage— No Cables - Complete Wireless Solution— Indicator LEDs – Visually Confirm Operation— Totally Sealed Case with Magnet On/Off— Remotely Managed and Configured via SMS,
MQTT explorer— Locally managed via Bluetooth Mobile
Application or USB attached PC Application— Real-time SMS Alerts, FTP and HTTP Uploads— Modular Communications(Bluetooth + 4G or Bluetooth + WiFi)— 4 x Internal D Cell Lithium Batteries

INSTALLATION— Low Power Battery Operation— Easy to install, no technician needed— Can Operate on External Power, No Batteries
installed— The description of how to install can befound at GNSS 5GV Quick Guide— Real-Time Clock synched to Cell Tower/ NTP— CSV/JSON/BIN Data Output,
Easy GISIntegration— Built-in Temperature Compensationwww.nti-audio.com

Mar 24Page 13

PRODUCT DATATechnical Specifications
Conformance

DIN 45669-1

Management Access PortUSB serial interface, Bluetooth (BLE)

Storage

Industrial MicroSD Card - 512 MB standard or 2 GB (optional upgrade)

Time Keeping

Real Time Clock (retains time for up to 3 months), Synched to NTP / CellTower

Vibration Limit (X, Y, Z)±40 g

Maximum Response

1 Hz to 1 KHz

Accuracy±2 %Sample Rate (Hz)

1000, 2000, 4000 samples/second

Reading interval

30 to 3600 sec

Peak Particle Velocity (by design)

0.003 mm/sec to 620 mm/sec

Peak Particle Velocity (validated)

Up to 620 mm/sec

Peak Ground Acceleration (by design)

0.003 gTime Stamping

Down to 1 millisecond

Buffer Size

8 MBAlerts

Thresholds: 1 to 150 mm/sec (@ 2G Range)
SMS Numbers: Can notify up to 5 mobile phones

Battery type4 x ER34615M - 3,6v /14,5 Ah (total 58 Ah) or 4 Saft LSH20D(total 52 Ah) - 3,6 v /13 Ah

Battery consumption•Up to 6 months in low-power mode - PPV/DF data is pushed throughthe 4G modem every 4 hours•

Up to 3 months in real-time MQTT mode - PPV/DF data is pushed
through the 4G modem in real time (every 30 seconds up to every 1hour depending on the user reporting interval set).

Structural monitoring

Peak amplitude/frequency values (1 Hz to 1 KHz) with Zero Crossing orFFT method, Peak Par­ticle Velocity (mm/s), Peak Vector Sum (mm/s).
Threshold alerts, and histori­c al trend readings.

Human response: (coming soon)

Acceleration 1/3 octave spectrum, vibration dose values (VDV) and acceleration total value Aw(t). Threshold alerts, and histori­c al trend readings.

www.nti-audio.com

Page 23

PRODUCT DATATechnical Specifications
Standards (frequency ranges)

ISEE seismograph, DIN 4150:3, DIN 4150:2, BS 7385, AS 2817.2-2006,ÖNORM S 9012, IN 1226, ICPE Circulaire 1986, NS 8176 COMFORT, NS8141 CONSTRUCTION, NS 8141-1, ISO 8569 ACC, SS 02211 SHAFT, SS4604861 COMFORT and SS 4604866 BLAST.

Standards (weightings)

ISO 2631:1989, ISO 2631:2003, UNI 9614-2017, BS6472-2008 and VC
curves

GSM Modems

EG25 (3G/4G) Global bands (uses nanoSIM)
BG96 (3G/4G) Global bands (uses microSIM)

Antennas

GSM, Bluetooth, GPS (optional)

Operating Temperature-40 °C TO +85 °C

Dimensions

W 152 x L 98 x D 41 mm

Ingress Protection

IP67

Weight1.15 kg (with battery base)

USB/ExternalPower 6-pin LEMO keyed connector with metal cap (IP68) and chainComes with••••Order Information NTi Audio #600 070 000

mounting bracket
USB cable
field carry bag
magnet and lanyard

Tilt Specifications
Tilt Readings

Pitch ± 90°, Roll ± 90°

Resolution

0.0035°

Accuracy+/- 0.005°

Alert Thresholds

0.05° to 70.0°

Temperature Stability+/- 0.005° (-45° to 85°)

Stabilisation Time

10 secs

Reading Interval

10 sec to 12 hours

All information is subject to change without notice.

www.nti-audio.com

Page 33