RIEGL
VZ-400i
800m
optional

1.2MHzThe RIEGL VZ-400i is a cutting-edge 3D Laser
Scanning System which combines a futureoriented, innovative new processing
architecture and internet connectivity
with RIEGL‘s latest waveform processing
LiDAR technology.
This real-time data flow is enabledthrough dual processing platforms:
a dedicated processing system fordata acquisition, waveform processing and system operations, anda second processing platform
which enables on-board dataregistration,
geo-referencing,
and analysis to be executed
simultaneously. Future improvements by continuous development will be available
via online firmware update.

RIEGL VZ-400i

Ultra High Performance 3D Laser Scanner
Redefining Productivity!
Typical Applications• Architecture & Facade Measurements • As-Built Surveying • Archeology & Cultural Heritage Documentation • City Modeling• Civil Engineering • Building Infrastructure Management (BIM) • Forensics & Crash Scene Investigation • Emergency Management• Tunnel Surveying • Forestry • Research • Monitoring

Scan this QR code towatch the VZ-400i video.

www.riegl.com

RIEGL LMS GmbH, Austria

RIEGL USA Inc.

RIEGL Japan Ltd.

RIEGL China Ltd.

RIEGL VZ-400iat a glance
RIEGL VZ-400i Main Features

VZ-400i Field Experience:

One of the fastest scanners on the market:
500+ scans (50 mdeg) within 8 hours,handled by one operator!

• ultra high speed data acquisition with up to 500,000 meas./sec,
survey-grade accuracy ≤ 5 mm, up to 800 m measurement range• easy to use / easy to train:user-friendly touchscreen interface, single touch operation, etc.
• orientation sensor for pose estimation• advanced flexibility through support for external peripherals and accessories,
e.g. external Bluetooth GNSS receiver on top• cloud connectivity via Wi-Fi and 3G/4G LTE• fully compatible with the RIEGL VMZ Hybrid Mobile Laser Mapping System• RiSCAN PRO standard processing software (included),RiSOLVE for fully automatic registration and colorization of scan data (optional)

RIEGL VZ-400i night scan in Vienna

Cloud Connectivity RIEGL VZ-400i

Cloud Storage(Data stored in VZ-400i project structure)
Compatible Cloud Storage Options:Amazon S3
FTP Server
Microsoft Azure

scan positions’ overview

Internet
3G/4G LTE
Wi-Fi

3G/4G LTE
Ethernet
Wi-Fidataemaile.g. .kml filee.g. .json filesmart phone

RIEGL VZ-400i in thefield transmits thedata after eachscan position

scanposition 5

scanposition 4

dataemaile.g. .kml filee.g. .json file

scanposition 1

scanposition 3

scan data detail,detail, reflectance-scaled

tabletesri

scan
position 2

scan position
overview,e.g. via ArcGIS Earth - esri

KML: Keyhole Markup Language (.kml file)
JSON: JavaScript Object Notation (.json file)

Further Application Examples:

laptopRIEGL VZ-400i Technical Data
800m max. measurement
range

1.2MHzoptional camera
optional

pulse repetition
rate PRR

onlinewaveformprocessing

multiple target
capability

Laser Class 1

Wi-Fi and
3G/4G LTE

construction
coonstruction site monitoring

Laser Pulse Repetition Rate PRR (peak)

100 kHz300 kHz600 kHz

1,200 kHz

Max. Effective Measurement Rate (meas./sec)

42,000125,000

250,000

500,000

Max. Measurement Range (≥ 90 %)

800 m

480 m

350 m

250 m

Max. Measurement Range (≥ 20 %)

400 m

230 m

160 m

120 m

Minimum Range

1.5 m

1.2 m

0.5 m

0.5 m

Accuracy / Precision

5 mm / 3 mm

Field of View (FOV)100° vertical / 360° horizontal
Laser Class 1 (eyesafe)

Eye Safety Class
Main Dimensions (width x height) / Weight

206 mm x 308 mm / 9.7 kg

forensics & investigation

Further details to be found on the current RIEGL VZ-400i Data Sheet.
RIEGL Laser Measurement Systems GmbH assumes no responsibility or liability what so everregarding the correctness, appropriateness, completeness, up-to-dateness, and quality
content and for the accuracy of the depicted objects respectively. All rights reserved.
© Copyright RIEGL Laser Measurement Systems GmbH, Horn, Austria

www.riegl.comPreliminary Info Sheet, RIEGL VZ-400i, 2016-09-29