This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12

DocumeentNumbeer:DocummentTitle:

DocumentInformationn
Revision
DocumentType

Vault
StatusDateInforrmation
EffectiveD
Date
ReleaseD
Date

NextReeviewDate
ExpirationDateControlInnformation
AuthorOwner

PreviouusNumber
ChangeeNumber

Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CONFIDENTIAL – This document is proprietary information and shall not be reproduced or disclosed
Rev D
without priorCPD-ELTF-1008
written approval.
AS-Series CSA Safety Test Report and Certificate (non-medical products)
EDF-004-010C
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010Form
1 12 revision: B

CPD-ELTF-1008 Rev D

AS-Series O2/Nitrogen Non-Medical CAN/CSA C22.2 61010-1-12/UL 61010-1 3rd Ed.
Project Name: AS Series Models
Design Owner: AirSep Corp.

PurposeThe purpose is to release the most recent safety testing for the AS Series models. This includes the report
and CSA certificate issued.
Summary of testing
CSA International
178 Rexdale Boulevard,
Toronto, ON, Canada M9W 1R3
Telephone: 416.747.4000 1.800.463.6727
Fax: 416.747.4149 www.csagroup.org
APPLICABLE REQUIREMENTS
CSA C22.2 No. 61010-1-12 - Safety requirements for electrical equipment for measurement, control, andlaboratory use — Part 1:
General requirements
UL 61010-1, 3rd Edition (May 11, 2012) - UL Standard for Safety Electrical Equipment For Measurement,
Control, and Laboratory
Use; Part 1: General Requirements - Third Edition
AS Series Models:
Oxygen Models
AS-A, AS-B, AS-D, AS-D+, AS-E, AS-G, AS-J, AS-K, AS-L,
AS-N, AS-P, AS-Q, AS-R, AS-W, AS-Z
Nitrogen Models
N2-AS-D+, N2-AS-E, N2-AS-G, N2-AS-J, N2-AS-K, N2-AS-L,
N2-AS-N, N2-AS-P, N2-AS-Q, N2-AS-Z

Voltage
110/120Frequency
60

Current (A)3.0240
Voltage
110/120240

50/60
Frequency
50/60
50/60

1.0
Current (A)3.0
1.0

Acceptance
The signed below have reviewed this test report and accept the results

APPROVED
Title
See Master Control Sign-offs
Name
Title

Title
Date:

Name
Title

Date:

Name
Title

Date:

Name
Title

Date:

Name

Date:

Name

Date:

EDF-004-010C rev B

Page 1 of 1
Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12

            





 

     

 
 
!
! "#$$  %&'()**    

 




 

     



 
 
  
 




           


 


              
               
 






 
 

           
   

   
 
 

+,*

+
, * -,   


   !"#
!"#!
#
!"#

!"#
!
#

#$%!#%!&#$%!#!#%!&! '(")
'("
( )
 *
*  +++ ,, -- ++  ..,/
,/
/ +
#$%!#%!&*#$%!#!#%!&! '(")
'("
( )
 *
   ++,0,
,0, ++ (1( 1++  )#!
#! #2)#!,./
,./ ,.
    !  ..!
!
  0
0 
3434+5+5 3#3#4
#4 6 .
.
.  
374
37
4

#1#8
#1$8
#198
#19:8#1 8 #1
#1;8
#1<8
#1=8
#1 8 #1
#1*8
#1)8
#1'8
#1!8
#1
#
#8
# #1
#
$8
$ #1
#
98 #1#
9:8 #1##
;8 #1#
<8 #1
#
=8
= #1
#
#
*8 #1#)8 #1#
'8 #1
#
!8
#1>8
#1?
#
#1
>8 #1
#
?

@ 


  + 3#4

#



A

 @

A

*+
*+2
+ 2
A 


 -B
B
B /
/
/      +8
++ 8 ..,/,// +   "8 ))+,
+, 9
9955  8 " ++,
++, +++5
+5
5  ""A

A 

 0  +,+, 2  ++,, AA'9  !B   C  1# ; A # ,5+  BA
Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.

)5This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12

   2



 

 
 2   A  2  2      
2

BB,,  // ++  ,+,,+, 22 * //22  ++  8 ////D8/D8 ++ E 
AA

)#!
)#!
#! $2" /
/ /   !  ..!
!
  0
0 
3434+5+5 3#3#4
#4 6 .
.
.  
374
37
4

*#1
# 9:8 *#1#  8 *#1# ;8 *#1# <8 *#1# =8= *#1#  8 *#1# *8 *#1# )8 **#19:8*#1
*#1;8*#1<8*#1=8*#1
*#1*8*#1)8#1'8*#1?
##1
'8 *#1
# ?

@ 


  + 3#4

#

 @

A

 @

A

*+
*+2
+ 2
A 


 -B
B
B /
/
/      +8
++ 8 ..,/,// +   "8 ))+,
+, 9
9955  8 " ++,
++, +++5
+5
5  ""A

A 

 0  +,+, 2  ++,, A
A/ ++
/2
//D8
AA
A BB,,  /  ,+,,+, 22 * /2  ++  8  ///D8 ++ E 

1
1  12

1 +23*+22"-
+ 3*
+2
3*+2
+22"
2 -
1# A *A     10+
10
0 +
+ .,., /./ ++ 0
0 
+
 ++ ,,

 .
.,/
.,/
/ + 0
0 // / // ++88  + 88   -- ++  
 F ) + 2;   .,;., /./ ++ 8   ,+,
10+( ,+, 3
3 8  4  ( 1+3+   0
0 10
0 +
+ +++ ,,
 .,/
.,/
/ + 6   // ++88  + 88   -- +
+ 
(G
,
(
(
G ) + 2 ;;  
 !.,! .
!., // ++  /,  ,+,,+,'9  !B   C  1# ; A # ,5+  BA
Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.

)5 This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12

   2



 

 
 2     2  2      
2*+2*++ 2) ++  ++,0,
,0,   34
 34  8   
B
B -  ++,0,
,0,   1#H "1%@"@    ,++,
,++, I,
,
I
I,+
,+ +
++ 1++    ,,, 0   314A II
IIIAA
I
IIAA

'9  !B   C  1# ; A # ,5+  BA
Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.

)5 This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12



 
     
 



 

      



          
     
 
 

                
             





 

 

  
        


!
! 
!



  

$%&      
 

     ""  
 !# $%$%&' ()"
,-
./
'
( )
)" * +(+( ,,- .//  00(( '    00  .!
! 
 
#
#



  
%&2      11 
 

   
  %
%&2 
 " "" 30""
6.4..


4"" 5 6
6
 '( ""  7 .4
4 ..   
 ""  88111  9 988         
:
:    8
8  
 ;   $ 



28#   
   // 2
8
8   
 ""  
"3"3 #



 ; 8 
 '&9  8
8  <<< =( > =( ? 
8 =#

  -@(( 
  # 2# ;    ! ; #    ""   
 -@
 . 
9"
 A# 9
9"  ""    (  (  (  ;(  (  (  1(
#
0 (   # $$    
  "" #    
 0
 


 ##

11

 1 8
 
  ""  8
8   (  *( 
 -( 
 $(
$ 
 6(
6 
 <( 
 B( 
 !(
! 
 2(
/
 C( 

 9(
9 
 D(
D  
  
 // )0)0  ;1

 1 (
D$29(   ,   
 " "" (   D$D$2 9
9( ""    1(
(  E  6 #  8  #Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.


  
    . 8 This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12

Descriptive Report

MASTER CONTRACT: 0000191105
REPORT: 1018431
PROJECT: 80204830
EDITION
EditionIssued On

Project

Location

Prepared By
John C.G. Marshall, P.
Eng.

Authorized By
John C.G. Marshall, P.
Eng.

70132743 RichmondSurbhi Seru, EIT

Surbhi Seru, EIT

80115139 Toronto

Wan Peng Han

Jean-Philippe Turcot

80133090 Toronto

Wan Peng Han

Dimcho Genov, P. Eng

CanadaTorontoHalley McLaren

Halley McLaren

Edition 1 December 16, 1988 1018431 - Rexdale
September 14,
2017
Edition 9 April 11, 2022
Edition November 28,
10
2022
EditionMay 8, 2024
11
Edition 880204830

CONTENTS
Report pages reissued, Attachments reissued
Contents
Certificate of Compliance + Supplement - Page 1 to 4
Description and Tests - Page 1 to 43
Att 1 - Rating Label – Pages 1 to 4
Att 2 - Photos and Illustrations – Pages 1 to 5
Attachments For CSA Engineering use only
Att 3 - Schematics – Pages 1 to 13
Att 4 - Device Manual – Pages 1 to 192
Att 5 - N2 Series Device Manual – Pages 1 to 55
Att 6 - IEC 61010-1 3rd Ed Checklist – Pages 1 to 89
PRODUCTS
C872106 ELECTRICAL LABORATORY EQUIPMENT - Electrical Laboratory Equipment
C872186 ELECTRICAL LABORATORY EQUIPMENT - Certified to US Standards
This report shall not be reproduced, except in full, without the approval of CSA Group.
The reader is responsible for any liability arising from actions taken in interpreting or applying the results presented in this report. This report shall not be
reproduced except in full, without written approval from CSA Group Testing & Certification Inc. The results of this report only relate to those items tested.

178 Rexdale
R d l Boulevard,
B ld Toronto,Tt ON,
ON Canada
C d M9W 1R3
Telephone: 416.747.4000 1.800.463.6727 Fax: 416.747.4149 www.csagroup.org
QD-1399 Rev 2023-05-19© 2024 CSA Group. All rights reserved.
Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 2
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
PART A:
Oxygen separation units, fixed, cord-connected
Model(s)AS-A, AS-B, AS-D, AS-D+, AS-E, AS-G, AS-J, AS-K, AS-L, AS-N,
AS-P, AS-Q, AS-R, AS-W, AS-Z

Voltage(VAC)

Frequency(Hz)

Current(A)

110/120

60

3.0

240

50/60

1.0

Notes:
1. The above models are cord connected, Equipment Class I, Pollution Degree 2, Installation Category II.
2. Mode of operation: Continuous.
3. Environmental Conditions: Normal: 4 to 40 C, 2000 m max, 0 to 80% rH.

PART B:
Nitrogen generator, fixed, cord-connected
Model(s)N2-AS-D+, N2-AS-E, N2-AS-G, N2-AS-J, N2-AS-K, N2-AS-L, N2AS-N, N2-AS-P, N2-AS-Q, N2-AS-Z

Voltage(VAC)

Frequency(Hz)

Current(A)

110/120

50/60

3.0

240

50/60

1.0

Notes:
1. The above models are cord connected, Equipment Class I, Pollution Degree 2, Installation Category II.
2. Mode of operation: Continuous.
3. Environmental Conditions: Normal: 4 to 40 C, 2000 m max, 0 to 80% rH.

APPLICABLE REQUIREMENTS
Standards Used
CSA C22.2 No. 61010-1-12

UL 61010-1, 3rd Edition (May 11, 2012)

Description
Safety requirements for electrical equipment formeasurement, control, and laboratory use — Part 1:
General requirements
UL Standard for Safety Electrical Equipment For
Measurement, Control, and Laboratory Use; Part 1:
General Requirements - Third Edition

MARKINGS
The manufacturer is required to apply the following markings:
x Products shall be marked with the markings specified by the particular product standard.
QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 3
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
x

Products certified for Canada shall have all Caution and Warning markings in both English and French.

Additional bilingual markings not covered by the product standard(s) may be required by the Authorities Having
Jurisdiction. It is the responsibility of the manufacturer to provide and apply these additional markings, whereapplicable, in accordance with the requirements of those authorities.
The products listed are eligible to bear the CSA Mark shown with adjacent indicators 'C' and 'US' for Canada andUS (indicating that products have been manufactured to the requirements of both Canadian and U.S. Standards) orwith adjacent indicator 'US' for US only or without either indicator for Canada only.
Nameplate adhesive label material approval information:
General Data M-714 label: 2 mil matte-chrome, polyester film with permanent emulsion acrylic adhesive andpolyester liner. UL approved (MH17205), Accepted for CSA.
The following markings appear on the product:
1. Submittor's identification (company name and/or file number: 191105 and/or registered tradename);
2. Marking on the unit that indicates the manufacturing location, if the equipment is manufactured at morethan one factory location;
3. Model designation;
4. Electrical rating;
5. Date of manufacture: Month and year of manufacture or date code. If a serial number is used instead ofdate of manufacture, a record of serial numbers shall be kept traceable to date of manufacture. (Not
related to date of sale);
6. The products listed are eligible to bear the CSA Mark shown with adjacent indicators 'C' and 'US' for
Canada and US (indicating that products have been manufactured to the requirements of both Canadianand U.S. Standards) or with adjacent indicator 'US' for US only or without either indicator for Canada
only:
,or
,or
.
7. Replacement fuse markings; Fuse type(s) and rating(s) (in volts and amperes), adjacent to thefuseholder(s).
8. Protective earthing TERMINAL is identified by the IEC 417 No 5019 symbol, adjacent to the
TERMINAL;
9. The IEC 417 Symbols 5007-a (“I”) and 5008-a ("O"), or equivalent, on or adjacent to the principal on/off
switch/circuit breaker when serving as a disconnect device; "I" inside a circle for push-push typeswitches.
10. Neutral is identified by the letter "N" (for units with a non-detachable supply cord).
The following additional markings are also provided:
1. The warning: "DISCONNECT SUPPLY BEFORE SERVICING".
DOCUMENTATION
Equipment is accompanied by the following documentation.
General: Technical specifications, instructions for use and details of where technical assistance may be obtained
if required.
QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 4
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830

Equipment Ratings: This includes equipment supply, description of I/O connections, duty cycle and operating
environmental conditions.
1.
2.
3.
4.
5.
6.
7.
8.

Pollution degree: 2.
Installation category: II.
Altitude: 2000 m.
Electrical supply: 240 Va.c., 50/60 Hz; or 110/120 Va.c., 60 Hz.
Indoor use statement.
Temperature: 4 to 40 °C.
Humidity: 0 to 80% rH.
Statement advising that mains supply voltage fluctuations are not to exceed 10 percent of the nominal
supply voltage.

Equipment Installation: This includes instructions for assembly and mounting, location requirements, details for
special services (cooling supply, etc.) as applicable.
Equipment Operation: This includes explanations of operating controls and warning symbols used, and instructions
for interconnection, replacement of consumables (e.g. paper) and cleaning as required.
Equipment Maintenance and Service: This includes instructions for preventative maintenance, inspection andcleaning, replacement of parts, etc.
1. Instructions on the following subjects shall be provided for service personnel, as necessary(NOTE: Instructions for service personnel need not be supplied, but should be made available to service
personnel).
ALTERATIONS
1. Markings as above.
2. Documentation as above.
3. Fuses (FU1, FU2) of 240V unit have alternate rating as 1.5A (LOCC item# 4.4c), 4.4d), 4.5f), 4.5g).

FACTORY TESTS
The following tests are conducted following production. Factory tests are conducted on 100% of
production.
Applicable
Factory Test
(indicate by
X)
X

QD-1399 Rev 2024-02-01

Type of Factory Test

Dielectric Voltage Withstand Test:(a)
Test voltages are raised to the specified value within 5 sec and maintained for 2 sec.
(b)
Test voltage is specified below for Installation Category II.
(c)
The factory test may be done at existing room temperature; equipment is not normallyenergized.

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 5
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Applicable
Factory Test
(indicate by
X)

Type of Factory Test

Warning: The factory test(s) specified may present a hazard of injury to personnel and/or
property and should only be performed by persons knowledgeable of such hazards and underconditions designed to minimize the possibility of injury.
For units up to 150V:
The equipment at the conclusion of manufacture, before shipment, shall withstand without
breakdown, the application of 840V ac or 1200V dc between mains TERMINALS and
ACCESSIBLE conductive parts.
For units rated above 150V and up to 300V:
The equipment at the conclusion of manufacture, before shipment, shall withstand without
breakdown, the application of 1400V ac or 2000V dc between mains TERMINALS and
ACCESSIBLE conductive parts.
X

Protective Earth Continuity Test:
Each unit that has a power supply cord with earthing conductor shall be tested to determine thatearthing continuity is provided between the PROTECTIVE (earthing) CONDUCTORTERMINAL and the ACCESSIBLE conductive parts of the unit that are required to be connectedto the PROTECTIVE CONDUCTOR TERMINAL.
Any indicating device (an ohmmeter, a battery and buzzer combination or the like) can be used todetermine compliance.

SPECIAL INSTRUCTIONS FOR FIELD SERVICES
1.

Component descriptions marked with either the “(INT)” or “(INT*)” identifiers may be substituted withother components providing the requirements specified under the notes in the “Description” are complied
with.

COMPONENT SPECIAL PICKUP
1.

Component descriptions marked with the identifier “(CT)” are subject to annual pickup and Conformity
Testing.

DESCRIPTION

Notes:
1. Component Substitutiona) Critical components (those identified by mfr name, cat no), which are NOT identified with either"INT" or "INT*" are not eligible for substitution without evaluation and report updatingb) The term "INT" means a "Certified" and/or "Listed" (or a "Recognized" and/or "Accepted")
component may be replaced by one "Certified" and/or "Listed" by another certification organization
accredited by the appropriate accreditation body or scheme requirements to the correct standard, forthe same application; providing the applicable country identifiers are included and requirements initem "d" below are complied with.
QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 6
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830c) The Term "(INT*)" means a "Recognized" and/or "Accepted" component may be replaced by acomponent that is CSA Certified. The applicable country identifiers shall be included, therequirements in item "d" below as well as any "conditions of suitability" for the component (asrecorded in this descriptive report) shall be complied with;d) Components which have been substituted, must be of an equivalent rating, configuration (size,
orientation, mounting) and the applicable minimum creepage and clearance distances are to be
maintained from live parts to bonded metal parts and secondary parts.
e) Substitution of a "Certified" and/or "Listed" component with a component that is "Recognized" or"Accepted" is not permitted without evaluation and report updating.
f) Substitution of a "Recognized" and/or "Accepted" component by one that is not CSA Certified is not
permitted without a proper evaluation as well as a report update because the Conditions ofAcceptance of the original component may be different than the Conditions of Acceptance of thesubstitute component.
General: These are cord connected, oxygen separation units and nitrogen concentration units. All models are
essentially similar in construction (such as with or without decorative enclosure which contain no electrical parts),
and vary in size of enclosure, size of molecular sieve beds and size of valves.
The ground-bond means is established as follows: The earthing conductor of the supply cord is provided with a
Certified/Recognized ring type pressure connector, which is secured to the metallic enclosure by dedicated, No. 6screw stud, star toothed washer and nut. The grounding section of the terminal block is connected by a separate
18AWG conductor, with a Certified/Recognized ring type pressure connector, which is attached to the stud with astar toothed washer and nut.
1.
2.
3.
4.
5.
6.
7.
8.
9.

Class of Equipment: I
Pollution Degree: 2.
Degree of Mobility: Fixed.
Environmental Conditions: Standard.
Operating Conditions: Continuous.
Connection to Supply Mains: Non-detachable cord connected.
Operating Temperature: 4 to 40 °C
Dimensions (L x W x H): 229 x 295 x 559 cm / 90 x 116 x 220 in (max)Weight: 8,826.9 kg / 19,460 lbs (max)

QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 7
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
LIST OF SAFETY CRITICAL COMPONENTS:
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademarkType / model

1

Main Enclosure

Hoffman
Enclosures
Inc

Cat No:
CSD16208/SPL

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptanceMaterial:
CSA61010- Accepted
Min 1/16 in 1
thick painted
UL61010-1
sheet steel.
Dimensions:
AS-A: 11 in
by 30-3/4 inby 5 in
AS-B: 141/8 in by 34
in by 5 in
AS-D: 161/2 in by 341/4 in by 51/2 in
AS-D+, ASE, AS-G,
AS-J, AS-K,
AS-L, AS-N,
AS-P, AS-Q,
AS-R, ASW, AS-Z:
20 in by 16
in by 9 inCover:
Samematerial
secured bydraw-pulllatches andby screws.

Additional information:
The ground-bond means is established as follows: The earthing conductor of the supply cord is provided with a
Certified/Recognized ring type pressure connector, which is secured to the metallic enclosure by dedicated, No. 6screw stud, star toothed washer and nut. The grounding section of the terminal block is connected by a separate
18AWG conductor, with a Certified/Recognized ring type pressure connector, which is attached to the stud witha star toothed washer and nut.
2. Absorber Vessels: (Typically two vessels are provided per unit).
Copies of the tank drawings are attached to the engineering file copy of this report.

QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 8
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark2.1

Absorber Vessel formodel: AS-A (INT)

Tri-Metal
Industries

Cat No:A033187-2

2.2

Absorber Vessel formodel: AS-B (INT)

Tri-Metal
Industries

Drawing No:
A033187

2.3

Absorber Vessel formodel: AS-D (INT)

Steel FabDrawing No:
A11752 L & R

QD-1399 Rev 2024-02-01

Type / model

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptanceMaterial:
CSA61010- Accepted
Cold rolled 1
steel
UL61010-1
Dimensions:
4 in OD by
56 in long;
10 Ga head,
0.083 inthick shell.
Working
Pressure:
100 PSI
Material:
CSA61010- Accepted
Cold rolled 1
steel
UL61010-1
Dimensions:
6 in OD by
57 in long;
12 Ga head,
0.120 inthick shell.
Working
Pressure:
100 PSI
Material:
CSA61010- Accepted
SA 414-G
1
Carbon SteelUL61010-1
Dimensions:
8 in OD by
57 in long;
0.094 inthick headand shell
Capacity:
11.4 gallons
Working
Pressure:
200 PSI, "U"
Stamped© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 9Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

2.3.a

Alternative Absorber
BrunnerDrawing No: 30Vessel for model: AS-D Engineering 5310
(INT)

2.4

Absorber Vessel for
Steel Fabmodel: AS-D+ (BE168)(INT)

Drawing No:
A127492.5

Absorber Vessel formodel: AS-E (INT)

Drawing No.:
A11751

QD-1399 Rev 2024-02-01

Manufacturer/
trademark

Steel FabType / model

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptanceMaterial:
CSA61010- Accepted
Cold rolled 1steel.
UL61010-1
Dimensions:
8 in OD by
54 in long;
0.094 inthick headand shell.
Capacity:
11.2 gallons
Working
Pressure:
100 PSI
Material:
CSA61010- Accepted
SA 414-G
1
Carbon SteelUL61010-1
Dimensions:
8 in OD by
58 in long;
0.094 inthick headand shell
Capacity:
11.7 gallons
Working
Pressure:
125 PSI, "U"
StampedMaterial:
CSA61010- Accepted
SA 414-G
1
Carbon SteelUL61010-1
Dimensions:
12 in OD by
56 in long;
0.094 inthick headand shell.
Capacity:
25.9 gallons
Working
Pressure:
200 PSI© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 10
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

2.5.a

Alternative Absorber
BrunnerDrawing No: 30Vessel for model: AS-E Engineering 6613.
(INT)

2.6

Absorber Vessel formodel: AS-G (INT)

2.6.a

Alternative Absorber
BrunnerDrawing No: 30Vessel for model: AS-G Engineering 5308.
(INT)

QD-1399 Rev 2024-02-01

Manufacturer/
trademark

Steel FabType / model

Drawing No.:
A11837

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptanceMaterial:
CSA61010- Accepted
Cold rolled 1steel.
UL61010-1
Dimensions:
12 in OD by
56 in long;
0.094 inthick headand shell.
Capacity:
26.2 gallons
Working
Pressure:
100 PSI
Material:
CSA61010- Accepted
SA414-G
1
Carbon SteelUL61010-1
Dimensions:
14 in OD by
65 in long;
0.094 inthick headand shell
Capacity: 40
gallons
Working
Pressure:
200 PSI, "U"
StampedMaterial:
CSA61010- Accepted
Cold rolled 1steel.
UL61010-1
Dimensions:
14 in OD by
65 in long;
0.094 inthick headand shell.
Capacity:
41.4 gallons
Working
Pressure:
100 PSI© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 11
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark2.7

Absorber Vessel formodel: AS-J

Steel Fab

2.7.a

Alternative Absorber
Vessel for model: AS-J
(INT)

BrunnerDrawing No: 30Engineering 5019.

QD-1399 Rev 2024-02-01

Type / model

Drawing No.:
A11838

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptanceMaterial:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:
20 in OD by
66 in long;
0.109 in min.
heads and
0.159 innom. shell
Capacity: 80
gallons
Working
Pressure:
200 PSI, "U"
StampedMaterial:
CSA61010- Accepted
Cold rolled 1steel.
UL61010-1
Dimensions:
20 in OD by
68 in long;
0.094 inthick headand shell.
Capacity:
86.8 gallons
Working
Pressure:
100 PSI© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 12
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark2.8

Absorber Vessel formodel: AS-K (INT)

Steel FabDrawing No.:C102884 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:
24 in OD by
72-3/16 in
long; 0.131
in min.
heads, 0.159
in nom. shell
Capacity:
110 gallons
Working
Pressure:
200 PSI, "U"
Stamped2.9

Absorber Vessel formodel: AS-L (INT)

Steel FabDrawing . No.:C102971 L & R

Material:
CSA61010- Accepted
SA516-70
1
Heads;UL61010-1
SA414-G
Shell
Dimensions:
26 in OD by
80 in long;
0.133 in min.
heads, 0.151
in nom. shell
Capacity:
172 gallons
Working
Pressure:
125 PSI, "U"
Stamped

QD-1399 Rev 2024-02-01

Type / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 13
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark2.10

Absorber Vessel formodel: AS-N (BE027)(INT)

Steel FabDrawing . No.:C101993 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:
30 in OD by
91 in long;
0.177 inhead minthick; 0.142
in shell min
thick
Capacity:
240 gallons
Working
Pressure:
125 PSI, "U"
Stamped2.11

Absorber Vessel formodel: AS-P (BE029)(INT)

Steel FabDrawing . No.:C101829 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:
36 in OD by
92 in long;
0.251 inhead minthick; 0.173
in shell min
thick
Capacity:
379 gallons
Working
Pressure:
125 PSI, "U"
Stamped

QD-1399 Rev 2024-02-01

Type / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 14
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark2.12

Absorber Vessel formodel: AS-Q (BE166)(INT)

Steel FabDrawing . No.:
C104244 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:
36 in OD by
117 in long;
0.187 inhead minthick; 0.173
in shell min
thick
Capacity:
480 gallons
Working
Pressure:
125 PSI, "U"
Stamped2.13

Absorber Vessel formodel: AS-R (BE099)(INT)

Steel FabDrawing . No.:
C102193 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:42 in OD by
128 in long;
0.226 inhead minthick; 0.214
in shell min
thick
Capacity:
713 gallons
Working
Pressure:
125 PSI, "U"
Stamped

QD-1399 Rev 2024-02-01

Type / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 15
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark2.14

Absorber Vessel formodel: AS-W (BE157)(INT)

Steel FabDrawing . No.:C103534 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:42 in OD by
165 in long;
0.226 inhead minthick; 0.214
in shell min
thick
Capacity:
927 gallons
Working
Pressure:
125 PSI, "U"
Stamped2.15

Absorber Vessel formodel: AS-Z (BE199)(INT)

Steel FabDrawing . No.:
C104223 L & R

Material:
CSA61010- Accepted
SA414-G
1
Carbon Steel UL61010-1
Dimensions:44 in OD by
190 in long;
0.188 inhead minthick; 0.250
in shell min
thick
Capacity:
1133 gallons
Working
Pressure:
125 PSI, "U"
Stamped

QD-1399 Rev 2024-02-01

Type / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 16
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark3

Power Supply Cord
(INT)

Mega
Electronics

SJTOW
N/14750
H05VV-F
IEC
S0227

18-3 AWG UL 817
UR E217655
Rating: 60 CSA No.21°C, 300 V
Attachment
Plug:
Molded-on
parallel blade
groundingtype.
Strain Relief:
Certified.
Heyco CatNo SR-7K-2,
mounted in a
double "D"shaped holein theenclosure toprevent
rotation.
Alternate
Strain Relief:
Certified.
Appleton Cat
No CG1850.

Littelfuse

Cat No: 342858

Rating: 15 UL 4248-1 UR E14721
A, 250 Va.c.
CSA 4248.1 CSA File
Mounting:
7316
Stem
mounted in"D" shapedhole inenclosure.

Type / model

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptance

4. Over-current Protection Devices
4.1

Fuse Holder
For use with model(s)AS-A, AS-B, AS-D

QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 17
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.1.a

Alternative Fuse Holder Bussmann
For use with model(s)AS-A, AS-B, AS-D

4.2

Allen
Fuse Holder
Bradley
For use with model(s)AS-D+, AS-E, AS-G,
AS-J, AS-K, AS-L, ASN, AS-P, AS-Q, AS-R,
AS-W, AS-Z

4.2.a

Alternate Fuse Holder
Bussmann
For use with model(s)AS-D+, AS-E, AS-G,
AS-J, AS-K, AS-L, ASN, AS-P, AS-Q, AS-R,
AS-W, AS-Z

4.2.b

Allen
Alternate Fuse Holder
Bradley
For use with model(s)N2-AS-D+, N2-AS-E,
N2-AS-G, N2-AS-J, N2AS-K, N2-AS-L, N2AS-N, N2-AS-P, N2AS-Q, N2-AS-R, N2AS-Z

Cat No: 1492-FB

Rating: 30A, UL 512
UR E34648
600 Vac
CSA No. 39
Mounting:
DIN Rail

4.3

120V MAINS Fuse
Littlefuse(FU1)
Mains protection (3A) inmodels rated: 110/120
Va.c.
Description: For use withmodel(s) AS-A, AS-B,
AS-D, AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 312003

Rating: 3 A, UL 248-1
250 V
CSA
Type: 1/4 by
No.248-1
1-1/4 in glass
cartridge
fuse

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptance

Rating: 16 UL 4248-1 UR E14853
A, 250 Va.c.
CSA 4248.1 CSA File
Mounting:
47235
Stem
mounted in"D" shapedhole inenclosure
Cat No: 1492-H6 Rating:
UL 1059
UR E40735
Max. 15A,
CSA No.158
300 Va.c.,
300 Vd.c., or
500 Va.c.,
500 Vd.c.
Mounting:
Din Rail
Cat No: HTB-461 Rating: 20A UL 4248-1 UR E14853(UL) or 16A
CSA 4248.1 CSA File
(CSA), 250
47235
V.
Mounting:
Din Rail
Cat No: HTB44M© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

UL E10480

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 18
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Standard

Mark(s) ofconformity
evidence of
acceptance

4.3.a

Alternative 120V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(3A) in models rated:
110/120 Va.c.
Description: For use withmodel(s) AS-A, AS-B,
AS-D, AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-3

Rating: 3 A, UL 248-1
250 V
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge
fuse

UR E19180

4.3.b

Alternative 120V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(1.5A) in models rated:
110/120 Va.c.
Description: For use withmodel(s) N2-AS-D+, N2AS-E, N2-AS-G, N2-ASJ, N2-AS-K, N2-AS-L,
N2-AS-N, N2-AS-P, N2AS-Q, N2-AS-R, N2-ASZ

Cat No: FNQ-R1-1/2

Rating: 1.5 UL 248-4
A, 600 Vac
Type: CC
type

UL E4273
CSA File
53787

4.3.c

Alternative 120V
Allen
MAINS Circuit Breaker Bradley(CR1):
Note: Mains protection(3A) in models rated:
110/120 Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492SP1C030

Rating: 3.0 A, UL 1077
UR E65138
277 Va.c., 48 CSA C22.2
Vd.c., -5 to 40 No. 235°C

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 19
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.3.d

Alternative 120V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(3.0A) in models rated:
110/120 Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

FNQ-R-3Rating: 3.0A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

4.3.e

Alternative 120V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(3.0A) in models rated:
110/120 Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

LP-CC-3Rating: 3.0A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

4.4

240V MAINS Fuse
Bussmann(FU1):
Note: Mains protection(1A) in models rated: 240
Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-1

Rating: 1 A, UL 248-1
250V
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse

UR E19180

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 20
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.4.a

Alternative 240V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(1A) in models rated: 240
Va.c.
Description: For use withmodel(s) N2-AS-D+, N2AS-E, N2-AS-G, N2-ASJ, N2-AS-K, N2-AS-L,
N2-AS-N, N2-AS-P, N2AS-Q, N2-AS-R, N2-ASZ

Cat No: FNQ-R-1 Rating: 1 A,
600 Vac
Type: CC
type

4.4.b

Alternative 240V
Allen
MAINS Fuse (FU1):
Bradley
Note: Mains protection(1A) in models rated: 240
Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492SP1C010

Rating: 1.0 A, UL 1077
UR E65138
277 Va.c., 48 CSA C22.2
Vd.c., -5 to 40 No. 235
°C

4.4.c

Alternative 240V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(1.5A) in models rated:
240Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

FNQ-R-1-1/2

Rating: 1.5A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceUL 248-1
CSA
No.248-1

UR E19180

cURus
E4273

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 21
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.4.d

Alternative 240V
Bussmann
MAINS Fuse (FU1):
Note: Mains protection(1.5A) in models rated:
240Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

LP-CC-1-1/2

Rating: 1.5A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

4.5

Fuse (FU2)
Bussmann
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-3

Rating: 3A, UL 248-1
250V.
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse.

UR E19180

4.5.a

Alternative Circuit
Allen
Breaker (CR2):
Bradley
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492SP1C030

Rating: 3.0 A, UL 1077
UR E65138
277 Va.c.,
CSA C22.2
48Vd.c., -5 to No. 235
40 °C

4.5.b

Alternative Circuit
Allen
Breaker (CR2):
Bradley
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492-H

Rating: 3.0 A, UL 60947 UR E40735
500 Va.c., 48 CSA 60947
Vd.c., -5 to 40
°C

4.5.c

Alternative Fuse (FU2): Bussmann
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-3

Rating: 3 A, UL 248-1
250 V.
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceUR E19180

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 22
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.5.d

Alternative 120V
Bussmann
MAINS Fuse (FU2):
Note: Mains protection(3.0A) in Schneider
Electric PLC panels
rated: 110/120 Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

FNQ-R-3Rating: 3.0A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

4.5.e

Alternative 120V
Bussmann
MAINS Fuse (FU2):
Note: Mains protection(3.0A) in Schneider
Electric PLC panels
rated: 110/120 Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

LP-CC-3Rating: 3.0A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 23
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.5.f

Alternative 240 MAINS Bussmann
Fuse (FU2):
Note: Mains protection(1.5A) in Schneider
Electric PLC panels
rated: 240Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

FNQ-R-1-1/2

Rating: 1.5A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

4.5.g

Alternative 240 MAINS Bussmann
Fuse (FU2):
Note: Mains protection(1.5A) in Schneider
Electric PLC panels
rated: 240Va.c.
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z, N2AS-D+, N2-AS-E, N2AS-G, N2-AS-J, N2-ASK, N2-AS-L, N2-AS-N,
N2-AS-P, N2-ASQ, N2-AS-R, N2-ASZ

LP-CC-1-1/2

Rating: 1.5A UL 248
CSA
600Vac,IR 200kA Vac No.248
Class CC,
Time-delay,
Dimension:
38mm x10mm(dia)

cURus
E4273

4.6

Fuse (FU3):
Littlefuse
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 312002

Rating: 2 A, UL 248-1
250 V.
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse

UL E10480

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 24
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Standard

Mark(s) ofconformity
evidence of
acceptance

4.6.a.

Alternative Fuse (FU3): Bussmann
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-2

Rating: 2 A, UL 248-1
250 V.
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse

UR E19180

4.6.b

Alternative Circuit
Allen
Breaker (CR3):
Bradley
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492SP1C020

Rating: 2.0 A, UL 1077
UR E65138
277 Va.c., 48 CSA C22.2
Vd.c., -5 to 40 No. 235
°C

4.6.c

Alternative Circuit
Allen
Breaker (CR3):
Bradley
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492-H

Rating: 3.0 A, UL 60947 UR E40735
500 Va.c., 48 CSA 60947
Vd.c., -5 to 40
°C

4.6.d

Alternative Fuse (FU3): Bussmann
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-3

Rating: 3A, UL 248-1
250V.
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse.

UR E19180

4.7

Fuse (FU4):
Littlefuse
Note: Over-current
protection (0.5A) for
Controller (see item 9).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 312.500 Rating: 0.5 A, UL 248-1
250 V
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge

UL E10480

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 25
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.7.aStandard

Mark(s) ofconformity
evidence of
acceptance

Alternative Fuse (FU4): Bussmann
Note: Over-current
protection (0.5A) for
Controller (see item 9).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-1/2 Rating: 0.5 A, UL 248-1
250 V
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse

UR E19180

4.7.b

Alternative Circuit
Allen
Breaker (CR4):
Bradley
Note: Over-current
protection (0.5A) for
Controller (see item 9).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492SP1C005

4.8

Fuse (FU5):
Littlefuse
Note: Over-current
protection (0.5A) forHMI (see item 11.1).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 312.500 Rating: 0.5 A, UL 248-1
250 V
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge

UL E10480

4.8.a

Alternative Fuse (FU5): Bussmann
Note: Over-current
protection (0.5A) forHMI (see item 11.1).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: AGC-1/2 Rating: 0.5 A, UL 248-1
250 V
CSA
Type: 1/4 by No.248-1
1-1/4 in glass
cartridge fuse

UR E19180

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

Rating: 0.5 A, UL 1077
UR E65138
277 Va.c., 48 CSA C22.2
Vd.c., -5 to 40 No. 235°C© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 26
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

4.8.b

Alternative Circuit
Allen
Breaker (CR5):
Bradley
Note: Over-current
protection (0.5A) forHMI (see item 11.1).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 1492SP1C005

Rating: 0.5 A, UL 1077
cURus
277 Va.c., 48 CSA C22.2 E65138
Vd.c., -5 to 40 No. 235°C

5

EMI Filter (EMI1)

Delta

Cat No:02DBAG5

Rating:
UL 1283
115/250V,CSA No.8
50/60Hz, 2A
Mounting:
Secured byscrews inside
enclosure

cURus
E79109

5.a

EMI Filter (EMI1)

CorcomCat No: 2VB1

Rating: 2 A, UL 1283
250 V, 50-60 CSA No.8
Hz, -10 to 40°C
Mounting:
Secured byscrews inside
enclosurecURus
E485705.b

EMI Filter (EMI1)

CorcomCat No: 3VB1

Rating:
UL 1283
3A, 250 V,
CSA No.8
50-60 Hz,
-10 to 40 °C
Mounting:
Secured byscrews inside
enclosurecURus
E48570

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 27
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

6

MAINS Transformer
SignalType: A41-175Transformer 230(TR1):
Description: For use with Comodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

6.a

Alternative MAINS
Hammond
Type: 185G230
Manufacturin
Transformer (TR1):
Description: For use with gmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

6.b

Alternative MAINS
Hammond
Power
Transformer (TR1):
Description: For use with Solutions
model(s) AS-D+, AS-E, Inc.
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

6.c

Alternate Mains
Transformer (TR1)

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkHammondPower
Solutions
Inc.

Type / model

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptance

Rating: Pri
CSA-C22.2(dual) 115/230 No.66.1
V, 1.52/0.76 UL506
A, 50/60 Hz,
175 VA, Sec(parallel) 115
V, 1.52 A
Insulation
Class: Class F(155°C)
Rating: Pri
UL 506(dual) 115/230
CSA-C22.2
V, 50/60 Hz,
No.66.1
175 VA, Sec(parallel) 115
V, 1.52 A
Insulation:
Class F (155
°C)

CSA
(221070),
cURus(E63829)

Type: 194263

Rating: Pri
UL5085-1(dual) 115/230 UL5085-2
V, 50/60 Hz, CSA No.
1.52/0.76 A, 66.1
175 VA, Sec
CSA No.
(parallel) 115
66.2
V, 175VA,
1.52 A
Insulation:
Class F (155
°C)

cULus
E50394

PH250PP

Rating:Primary:
120/240V
250VA
50/60Hz
Secondary:
120/240V
250VA
Isolation
Class B

cULus
E50394© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

UL5085-1
UL5085-2
CSA No.
66.1
CSA No.
66.2

cURus(E207860)

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 28
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademarkType / model

7

Optional Transformer
ATC Frost
Magnetics /(TR2):
StandexNote: For use with
Optional Alarm Board(see item 20).
Description: For use withmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Type: FT2432

8

Controller (INT)

8.a

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptance

Rating: Pri
UL506
115 V, 50/60
Hz, Sec (dual)
5/10 V,
1.2/0.1A
Insulation:
Class B (130°C), Thermal
Fuse: 115 °C

UR E105782

Omron Corp Type: ZEN10C2AR-A

Rating: 100 – UL 508
240 Va.c.,
CSA No.
50/60 Hz, 30 142
VA max, -25
to 55 °C

UR E95399

Alternative Controller(INT)

Omron Corp Type: ZEN20C2AR-A-VZ

Rating: 100 – UL 508
240 Va.c.,
CSA No.
50/60 Hz, 23 142
VA max, -25
to 55 °C

UR E95399

8.b

Alternative Controller
(INT)

Allen
Bradley

8.c

Alternative Controller(INT)

Omron Corp Type: ZEN20C2AR-A-V2

Rating: 100 – UL 508
240 Va.c.,
CSA No.
50/60 Hz, 7
142
VA max (for
100 Va.c.), 10
VA max (for240 Va.c.), -25
to 55 °C

UR E95399

8.d

Alternative Controller(INT)

SchneiderElectric

Rating: 85-264 UL 508
Vac, 50/60Hz,
58VA max, 10 to 55 °C

UR E468248

QD-1399 Rev 2024-02-01

Type: Micrologix Rating: 85 – CSA C22.2 UR
265 Va.c., 47- No. 142
1200
63 Hz, 70 VA Class I Div2
max, 0 to 55 UL 508°C

Type: Modicon
M221© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 29
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark8.e

Alternative Controller
Schneider(INT) Description: For Electricuse in model(s) AS-B,
AS-D, AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N & AS-P

Type / model

Technical data

Type: Zelio SR2
E201FU

Rating: 85264VAC,
50/60Hz, 11
VA max, -20
to 40 °C

Standard

UL 508
CSA No.
142

Mark(s) ofconformity
evidence of
acceptancecULus
E316433

9. Manual Interface Components
9.1

Printed Wiring Board:
Description: For use inmodel(s) AS-A, AS-B,
AS-D

4FrontSolutionsLLC

9.2

Printed Wiring Board
Connectors: (INT)
Description: For use inmodel(s) AS-A, AS-B,
AS-D

Panduit

QD-1399 Rev 2024-02-01

Material: 1/16 UL94
in thickfiberglass
epoxy board
Dimensions: 4
in by 4 in.
Function:
Mountscomponents in
low and extralow voltage
circuits.
Spacings:
0 to 30V:
0.04mmmaintained intransient
limited circuits
per C22.2 No
142-M1987.
120V:
1.05mmmaintained per
C22.2 No 142M1987.
CONN RCPT UL94
Cat No:CE156F22-15 and 15P IDC
22AWG TINMLSR156-15.
LEAD
CB144-1© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

URURThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 30
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark9.3.1

On/Off Switch
Description: For use inmodel(s) AS-A, AS-B,
AS-D

Carling
Cat No:technologies LT1GK61

9.3.2

On/Off Switch
Idec Corp
Description: For use inmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: ASLD- Rating: 6 A, UL508
UR E68961
200 operator,
120 Va.c.
CSA C22-2
BST-010, or BST- Type: Panel No.14001 contact block. mounted,
lighted, SPST
rotary switch,
consisting ofoperator andcontact block.

9.3.3

Allen
On/Off Switch
Description: For use in Bradleymodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 800FM
series

9.4.1

Auto/Manual Switch:
Description: For use inmodel(s) AS-A, AS-B,
AS-D

QD-1399 Rev 2024-02-01

Type / model

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptance

Rating: 6 A, UL 1054
UR E7560
125 Va.c.
UL 61058-1
Type: Panel CSA No.55
mounted,CSA 61058lighted DPST
1
rocker switch.
Mounting:
Secured in
enclosureopening.

Rating: 10 A, UL 508A
UR E14840
690 V,CSA C22.2–25 to 70°C
No. 14
Type: Panelmounted,lighted, SPST
rotary switch,
consisting ofoperator andcontact block.

Carling
Cat No: LTILA61 Rating: 6 A, UL 1054
UR E7560
technologies.
120 Va.c.
UL 61058-1
Type: Panel CSA No.55
mounted,CSA 61058lighted SPST
1
rocker switch.
Mounted:Secured in
enclosureopening.

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 31
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark9.4.2

Auto/Manual Switch:
Allen
cULus
Bradley
Description: For use inmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 800FM
seriesRating: 10 A, UL 508A
UR E14840
690 V, –25 to CSA C22.2
70 °C
No. 14
Type: Panelmounted,lighted, SPST
rotary switch,
consisting ofoperator andcontact block.

9.5.1

Allen
Pushbutton Switch:
Description: For use in Bradleymodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 800FM
seriesRating: 10 A, UL 508A
UR E14840
690V, –25 to CSA C22.2
70°C
No. 14
Type: Panelmounted,lighted, SPST
rotary switch,
consisting ofoperator andcontact block

9.6.1

Hour Meter

Hobbs

Cat No: 20184

Rating: 120 UL
V, 60 Hz, 3 W CSA
max
Mounting:
Screws

UL, CSA

9.6.2

Hour Meter:
SendecDescription: For use inmodel(s) AS-B, AS-D,
AS-D+, AS-E, AS-G,
AS-J, AS-K, AS-L, ASN, AS-P, AS-Q, AS-R,
AS-W, AS-Z

Cat No: 807D800-8111

Rating: 120 UL
V, 60 Hz, 3 W CSA
max
Mounting:
Press inretaining clips

cETLus9.6.3

Hour Meter: Description: Sendec
For use in model(s) ASB, AS-D, AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: N3300200

Rating: 120
V, 60 Hz,
150uA maxload current
Mounting:
Integrated
snap-in case

cETLusType / model

Technical data

Standard

UL
CSA

Mark(s) ofconformity
evidence of
acceptance

10 Optional Electronic Interface Components:
Description: Alternative user interface replaces all components in Manual Interface (see item 9).
QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 32
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

10.1

Allen
HMI: (INT)
Description: For use in Bradleymodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Type: PanelView Rating: 85 –264 VAC, 47
600
Cat No: 2711P- – 63 Hz, 60
VA max, 0 –
T6M5A
55 °C

ANSI/ISA UR E10314
12.12.01
CSA
No.213
Enclosure:
UL 50
CSA C22.2
No. 94.2-07

10.2

Schneider
HMI:
Description: For use in Electricmodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Type: Magelis
STO & STU
Cat No:HMISTU855

UL 508UR E164866
Ul 61010-1
CSA
No.142
CSA 610101

11

Solenoid Valves

Automatic
Switch Co

Cat No: series
Rating: 120 UL 429
CSA File
8256, 8210, 8262, V, 50/60 Hz, CSA No.139 010381_C_00
11 to 16.7 W,
8225, 1890150psi

11.a

Alternative Solenoid
ValvesASCOCat No:18900036

Rating:CSA File
UL 429120/110 V,
CSA No.139 010381_C_00
50/60 Hz, 2.50W, 150 psi

11.b

Alternative Solenoid
ValvesASCOCat No:8210G054

Rating: 120V, UL 429
CSA File
60 Hz, 16.1
CSA No.139 010381_C_00
W, 150 psi.
011.c

Alternative Solenoid
ValvesASCOCat No:8290A394

Rating: 120V, UL 429
CSA File
60 Hz, 150 psi. CSA No.139 010381_C_00
011.d

Alternative Solenoid
ValvesASCOCat No: Series
8551

Rating:CSA File
UL 429120/110V,
CSA No.139 010381_C_00
50/60 Hz,
0
2.5W, 150psi
max

QD-1399 Rev 2024-02-01

Manufacturer/
trademarkType / model

Technical data

Rating: 20.428.8 VDC,
6.8W, 0 – 50
°C

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptanceThis copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 33
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark11.e

Alternative Solenoid
ValvesGranzow

Cat No: ET20M

12.1

Pressure Switch

Automatic
Switch Co

Cat No:Rating: 5A
UL
CSA, UR
PA16A/RF10A71 (res.),
Recognized
125/250V ac. CSA
10-100,200psi Certified

12.2

Allen
Pressure Switch
Description: For use in Bradleymodel(s) AS-D+, AS-E,
AS-G, AS-J, AS-K, ASL, AS-N, AS-P, AS-Q,
AS-R, AS-W, AS-Z

Cat No: 836 series Rating: 5A
250V ac.
-30-375,
750psi

12.3

Pressure Switch
Description: For use inmodel(s) AS-A, AS-B,
AS-D

Cat No: PG16A

13

Cat No: SV25,
Rating: 100
Pressure Relief Device Control
Devices Inc ST-25, ST-33, SF- PSI, 150 PSI
ALL set to 100psi
50, SB-50, SW10
AS_D+,E,G,J,K,N,L,P,Q,
B, D, Q,R,W,Z

ASME

ASME
Certificate
14396

13.1

Alternative Pressure
KunkleRelief Device
AS_D+,E,G,J,K,N,L,P,Q,
B, D, Q,R,W,Z

Cat No:6000series

ASME

ASME
Certified

14

Connectors (INT)

Type: Crimp-on, 18A, 600V
UL
insulated push-on max, 18AWG,
type
105 C

ASCO

AMP Inc

Type / model

Technical data

Rating:120VAC,
50/60 Hz20W
300 psig

Standard

Mark(s) ofconformity
evidence of
acceptance

CSA 61010- Acceptedwith project1
UL 61010-1 70132743

UL 508UL E114842
CSA No. 14 CSA File
LR1234

Rating: 5A
UL
CSA, UR
250V ac.
Recognized
10-100,200psi CSA
CertifiedRating: 100
PSI

UL E326725

12A, 600V
max, 105 C, 22
AWG
15Internal Wire (INT)

QD-1399 Rev 2024-02-01

Alan Wire
Co

Type: TEW

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

UL 1063

UR E83039

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 34
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademarkType / model

ColemanCable Inc

Technical data

Standard

Mark(s) ofconformity
evidence of
acceptance

Rating: 600V,
UR E137733
22AWG, 18
UL
AWG, 105°C Traceability
Requiremen UR E129971
ts

Omni Cable
Corp

UL
Traceability
Requiremen
ts
16

Alternate Wire Harness: OLFLEX
Description: For use inmodel(s) AS-R, AS-W,
AS-Z

Type:601603

17

Optional Alarm Board
Fujikura(Oxygen Sensor Module) Japan

CSA61010- Accepted
FCX-MWAS-ST1 Rating: 5
VAC, 101
VAC, 0 to 60 UL61010-1
°C

17.1

Alternative Optional
Alarm Board (Oxygen
Sensor Module)

Airsep

CB130-1

Rating:
CSA61010- Accepted
115VAC, 4 to 1
40 °C
UL61010-1

17.2

Alternative Optional
Alarm Board (Oxygen
Sensor Module)

Advanced
Micro
Instruments
Inc

Model 65

Rating: 7CSA61010- Accepted
28VDC, 0 to 1
130 °F,UL61010-1
250mA at 12V

Rating 600V, CSA61010- Accepted
18AWG, 90°C 1
UL61010-1

18. Optional Quick Connectors:
Note: Used for connections to solenoid valves, pressure switches and pressure relief devices only.
18.1

Optional Quick
Connector (female
receptacle)

Hubbell Inc

MRMS24325

Rating: 300 UL 2238
CSA (LR
VAC, 300
CSA C22.2 285), cULus
VDC, 4 A, -20 No.182.3(E192071)
to 105 °C,
1/4in NPT

18.2

Optional Quick
Connector (male plug)

Hubbell Inc

MPMS2415

Rating: 300 UL 2238
CSA (LR
VAC, 300
CSA C22.2 285), cULus
VDC, 4 A, -20 No.182.3(E192071)
to 105 °C

QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 35
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Uniquecomponent
referenceor
location

Application / function

Manufacturer/
trademark18.3

Optional Quick
Connector (male plug)

ASCO

400125-098

Rating: 115 CSA
VAC, 4 A, -20 UL
to 75 °C

19

Acoustic Signaling
Alarm
(INT)

Allen
Bradley855P-B10ME22

Rating: 120 UL 508
UR E14840
VAC, 42 mA, CSA C22.2
100 dB at 1
No. 14meter

20

Optional Power Supply Meanwell
Description: For use withdisplay and flowmeter

MDR-20-24

Rating: 85UL508
264VAC,
0.55A/115VA
C
0.35A/230VA
C, 47-63Hz, 20 to 85 °C

20a

Optional Power Supply Phoenix
Contact(Alternate)
Description: For use with
Schneider Electric PLC
panels

UNORating:
UL 60950-1 cURus
PS/1AC/24DC/30 100-240Vac CSA 60950- E214596
W
1
50-60Hz(2902991)
0.8A 100Vac/
0.4A 240Vac
Output:24Vdc
1.25A

Type / model

Technical data

Note:

QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Standard

Mark(s) ofconformity
evidence of
acceptance

UR, CSAUR E215312

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 36
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
TEST HISTORY
Edition 1 – Application LR 83232-1:
The following tests were conducted on model: AS-20.
1. Rating:
2. Temperature:
3. Dielectric Strength:
4. Leakage Current: (C22.2 No 68, Cl 6.10)
5. Hydrostatic Tests: (C22.2 No 68, Cl 6.18)
Edition 2 – Application LR 83232-4:
No testing was deemed necessary.

Edition 3 – Project 1018431:
The tests summarized below were conducted with satisfactory results. Detailed test results are on file at CSA.
Device Tested: Oxygen Separator, Model AS45
Test Requirements: CSA C22.2 No 1010.1/UL 3101.1.
Note: Checklist Location in the following table refers to the form for test results found in the CSA 1010.1
Conformance Verification Report.
Clause:4.4
4.4.1
4.4.2.1
4.4.2.2
4.4.2.3

Description:

Compliance:
Y N N/A
X
X
X
X
X

Checklist
location:
Form A.2
Form A.2
Form A.2
Form A.2
Form A.2X

Forms
A.2/A.25
Form A.2
Forms A.2/
A.27/A.28
Form A.2
Form A.2

4.4.2.4

Testing in SINGLE FAULT CONDITION
Non-specific faults
PROTECTIVE IMPEDANCE
Protective conductor (Earth leakage current)
Equipment or parts for short term orintermittent operations
Motors

4.4.2.5
4.4.2.6

Capacitors
Mains transformers

XX

4.4.2.7
4.4.2.8

Outputs
Equipment for more than one supply (also Cl.
14.5)
CoolingHeating devices
Insulation between circuits and parts
Interlocks
Mains Supply (Input)
Durability of markings
Determination of ACCESSIBLE parts:
General examination
Top openings
Preset control openings
Permissible limits for ACCESSIBLE parts:

XX

4.4.2.9
4.4.2.10
4.4.2.11
4.4.2.12
5.1.3
5.3
6.2
6.2.1
6.2.2
6.2.3
6.3

QD-1399 Rev 2024-02-01

X
X
X
X
X
X
X
X
X
X

Form A.2
Form A.2
Form A.2
Form A.2
Form A.3
Form A.4
Form A.6
Form A.6
Form A.6© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Comments:

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 37
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Clause:
6.3.1
6.3.1.1
6.3.1.2
6.3.1.3
6.3.2
6.3.2.1
6.3.2.2
6.3.2.3
6.5.1
6.5.1.2

Description:

7.4
8
8.1

Values in NORMAL CONDITION:- Voltage- Current- Capacitance
Values in SINGLE FAULT CONDITION:- Voltage- Current- CapacitanceProtective Earthing:
Bonding impedance, plug-connected
equipment
Bonding impedance, PERMANENTLY
CONNECTED EQUIPMENT
Indirect bonding from measuring and test
equipment
Separation of circuits - short circuits due toinsufficient separation
Terminals for external circuits - charge on
capacitors
Creepage and clearance distances:
Working voltage
Comparative tracking index (CTI)
10 N on bare conductor
30 N on outside enclosure
Voltage (dielectric strength) tests:
After humidity preconditioning
After enclosure tests (Cl. 8)
After resistance to heat (Cl. 9)
After SINGLE FAULTS (Cl. 4.4.4)
After cord entry (Cl. 6.10.2.1)
After protection against hazards from fluids(Cl. 11)
Fitting of non-detachable mains supply cords:
Cord entry
Cord anchorage
Capacitor discharge - NORMAL CONDITIONFree strand test
Stability:
10 deg tip
Horizontal force
Downward force
Provisions for lifting and carrying
Mechanical resistance to shock and impact:
Rigidity

8.2

Impact hammer test

8.4

Drop test:

6.5.1.3
6.5.1.4
6.6.1
6.6.2(6.1.1)
6.7

6.8

6.10.26.10.2.1
6.10.2.2
6.10.36.11.1
7.3

QD-1399 Rev 2024-02-01

Compliance:
Y N N/A

Checklist
location:

X
X
X

Form A.7
Form A.7
Form A.7

X

Form A.8
Form A.8
Form A.8

X
X
X

Form A.9
X

Form A.9

X

Form A.9

X

Form A.2X

Forms
A.6/A.7

X
X
X
X

Form A.5
Form A.5
Form A.11
Form A.11

X
X
X

Form A.12
Form A.12
Form A.12
Form A.12
Form A.12
Form A.20

X
X
X
X
X

Form A.13
Form A.13
Form A.6
6.11.1

X

7.3
7.3
7.3
7.4

X
X
X
X
X

X
X
X

Forms
A.11/A.20
Forms
A.11/A.20© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Comments:

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 38
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
Clause:Description:

8.4.1

Equipment other than hand-held

8.4.2

Hand-held equipment

9.2
9.2.1
9.2.2

10.3
11
11.2
11.3
11.4
11.5
11.6
11.7
12.2.1
12.3

Temperature, general
Temperature, heating equipment
Temperature, equipment intended for
installation into a cabinet or wall
GuardsResistance to heat of non-metallic
ENCLOSURESVICAT softening test (ISO 306)
Protection against hazards from fluids:
Cleaning
Spillage
Overflow
Battery electrolyte
Specially protected equipment (IEC 529)
Fluid pressure and leakage
Ionizing Radiation (IEC 405)
Ultra-violet radiation

12.4

9.3
10.2

Compliance:
Y N N/A
X

X
X

Checklist
location:
Forms
A.11/A.20
Forms
A.11/A.20
Form A.18
Form A.18
Form A.18

X
X

Form A.18
Form A.19

X

10.3

X
X

X
X

Form A.20
Form A.20
Form A.20
11.5
Form A.20
Form A.21
Form A.22
N/A

Microwave radiation

X

N/A

12.5.1
12.5.2

Sound level
Ultrasonic pressure (max. 22.4 kHz)X
X

Form A.23
Form A.23

12.6
13.1
13.2.213.3
14.2.1
14.3
14.4
14.5
14.6

Laser sources (IEC 825)
Poisonous and injurious gases
Batteries
Implosion of high-vacuum devices (IEC 65)
Motor temperatures (also 4.4.2.4)
Over-temperature protection devices
Fuseholder accessibility
Mains voltage selecting devices (also 4.4.2.8)
High integrity components (appropriate IEC
publications)
Mains transformers; short circuit
Mains transformers; overload
Interlocks
Protection against the spread of fire:
Limited circuits
Flammability

X
X
X
X
X
X
X
X
X

12.6
attach data
Form A.24
13.3
Form A.25
Form A.26
14.4
Form A.2
Form A.10

X
X
X

Form A.27
Form A.28
15

X
X

Form A.15
Form A.16

14.7.1
14.7.215
Annex F
F.2.1
F.4.3.3

QD-1399 Rev 2024-02-01

X
X
X
X
X
X

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Comments:

Test underconsideration.
Test underconsideration.
Test underconsideration.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 39
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
CANADIAN and U.S. SUPPLEMENT TO IEC 1010-1
Clause:6.7
6.10.4.5
6.10.4.5.1
6.10.4.5.2
6.10.4.5.3
6.10.4.5.4
13.4
13.4.113.4.213.4.313.4.414.8.1

Description:
PCB Coatings
Conduit Enclosure Entry Tests
Conduit Pull-out
Conduit Torque Test
Bending
Knockouts
Fluid Pressure and Leakage
Hydraulic Test Pressure
Maximum Permissible Working Pressure
Pressure Relief Devices
Leakage Test
Conductive Coatings

Compliance:
Y N N/A
X
X
X
X
X
X
X
X
X
X
X
X

Checklist
location:

Comments:
CSA Certified

Edition 4 – Project 1457595:
The tests summarized below were conducted with satisfactory results on a representative sample of Mains
Transformer (TR1); Manufacturer: Signal; Type: A41-175-230. Detailed test results are on file at CSA.
Clause:14.7.1
14.7.2

Description:
Mains transformers; short circuit
Mains transformers; overload

Compliance:
Y N N/AX
X

Checklist
location:

Comments:

Edition 5 – Project 1577889:
No test was deemed necessary.
Edition 6 – Project 2342553:
Updated to the latest edition of the standard. Product was found to be in compliance with the applicablerequirements. The following models are added: AS-D+, AS-P, and AS-W. The following models are removed:
AS-20, AS-45, AS-80, AS-160, AS-250, AS-450, AS-750, AS-1000, and AS-1250.
The tests summarized below were conducted with satisfactory results on a representative sample of model: AS-E.
Detailed test results are available in Attachment 4. Testing was conducted in Cleveland, OH.
Compliance Criteria: Y = Pass, N = Did Not Pass, N/A = Not Applicable
Clause:Description:

4.4.2.6.1
5.1.3
6.5.1.3
6.8
10
Annex G.3

Mains transformers - Short circuit
Mains Supply
Bonding impedance of plug connected equipment
Dielectric Strength Tests:
Temperature Measurements
Leakage and Rupture from Fluids Under Pressure

QD-1399 Rev 2024-02-01

Checklist
Location:
Form A.2
Form A.3
Form A.10
Form A.14
Form A.20A
Form A.24© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

Compliance:(Y, N, N/A)
Y
Y
Y
Y
Y
Y

Comments:-------

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 40
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830

Edition 7 – Project 2591581:
Update report to address FIR findings dated 2011-06-08, 2011-10-19, and 2012-08-22. List of Safety Critical
Components was updated to address the FIR findings. No additional testing was deemed necessary.
Edition: 8 (Project 70132743)
Report updated to upgrade to 61010-1 3rd Edition and cover cCSAus Certification of new Nitrogen Concentrator
model series N2-AS.
The tests summarized below were conducted with satisfactory results on a representative sample of model N2AS-G configured for the worst conditions of normal operation and representative of both Part A and Part B
models. Detailed test results are on file at CSA.
Device Tested: Nitrogen Concentrator N2-AS-G
Test Location: CSA Group Testing and Certification Inc.,
13799 Commerce Parkway, Richmond, BC, Canada V6V 2N9and
CSA Group Testing and Certification Inc.,
178 Rexdale Blvd., Toronto, ON, Canada M9W 1R3
Test Requirements:
CSA C22.2 No 61010-1-12 and ANSI/UL 61010-1:2012
Note: Checklist Location in the following table refers to the form for test results found in the CSA 61010-1-12
Checklist
IEC 61010-1-10 Check List:
Clause:Description:

4.4.2.3
4.4.2.7.2
4.4.2.7.3
5.1.3
5.3
6.2
6.3.1

Protective conductor
Mains transformers – short circuit
Mains transformers – overload
Mains supply
Durability of markings
List of accessible parts
Limit values for accessible parts – levels in normal
condition
Limit values for accessible parts – levels in single fault
condition
Bonding impedance of plug connected equipment
Insulation requirements – Clearances and Creepages
Dielectric strength tests
Cord anchorage
Sharp Edges
Enclosure rigidity
Protection against the spread of fire
Temperature Measurements

6.3.2
6.5.2.4
6.7
6.8
6.10
7.2
8.2
9
10

QD-1399 Rev 2024-02-01

Checklist
Location:
Form A.6
Form A.39
Form A.40
Form A.2
Form A.3
Form A.4
Form A.5

Compliance:(Y, N, N/A)
Y
Y
Y
Y
Y
Y
Y

Form A.6

Y

Form A.9
Form A.15
Form A.18
Form A.19
Form A.43
Form A.21A
Form A.22
Form A.26

Y
Y
Y
Y
Y
Y
Y
Y

© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 41
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830

The tests summarized below were conducted with satisfactory results on a representative sample of model the
Granzow ET20M solenoid valve. Detailed test results are on file at CSA under project 70132743.
Device Tested: Granzow ET20M solenoid valve
Test Location: CSA Group Testing and Certification Inc.,
178 Rexdale Blvd., Toronto, ON, Canada M9W 1R3
Test Requirements:
CAN/CSA C22.2 No. 139-13 and ANSI/UL 429:2013
CAN/CSA C22.2 No. 139-13 and ANSI/UL 429:2013:
Clause:7.10/sec 34
7.18/sec 39
7.6/sec 37

QD-1399 Rev 2024-02-01

Description:
Strength tests of fluid-containing parts
Burnout test
Dielectric strength

Compliance:(Y, N, N/A)YYY© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 42
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830

Edition 9: Project 80115139
Report updated to include the alternative transformer provided by Hammond Power Solutions, Model.PH250PP.
LOCC updated with Table format as per CSA Descriptive Report required.
Change is considered as replacement of an existing part for a new part with same ratings and approvals and is inline with Measurement, Control and Laboratory Equipment Certification Informs No. 26
The tests summarized below were conducted with satisfactory results on a sample of model PH250PP fortransformer testing outside the equipment. Detailed test results are on file at CSA.
Device Tested: Transformer PH250PP
Test Location: CSA Group Testing and Certification Inc.,
178 Rexdale Blvd., Toronto, ON, Canada M9W 1R3
Test Requirements:

CSA C22.2 No 61010-1-12 and ANSI/UL 61010-1:2012

Test performed:

4.4.2.7 Transformer (OL/SC)
10.2 Temperature of winding
14.6 Mains transformer tested outside equipment

Construction Review:Construction review performed with satisfactory results.
Edition 10: Project 80133090
Report updates to add alternate component to LOCC:- Item 4.3, FU1 for 120V unit. (3.0A)- Item 4.4, FU1 for 240V unit. (1.5A)- Item 4.5, FU2 for 120V unit. (3.0A)- Item 4.5, FU2 for 240V unit. (1.5A)- Item 5, EMI filter (EMI1) for 120V unit.
- Item 20, alternate PSU. (For use with Schneider Electric PLC panels display and flowmeter only)
The tests summarized below were conducted with satisfactory results on an end unit sample of model AS-D+.
Detailed test results are on file at CSA.
Device Tested: Oxygen separation units, Model AS-D+
Test Location: CSA Group Testing and Certification Inc.,
178 Rexdale Blvd., Toronto, ON, Canada M9W 1R3
Test Requirements:

CSA C22.2 No 61010-1-12 and ANSI/UL 61010-1:2012

Test performed:
4.4.2.7 Transformer (OL/SC)
QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series
Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12
MASTER CONTRACT:
0000191105

Page No: 43
Date Issued: May 8, 2024

REPORT: 1018431
PROJECT: 80204830
5.1.3 Power Input
6.3.1 Permissible Limits - Normal condition
6.3.2 Permissible Limits – Single fault condition
10.1 / 10.3 Temperature
10.2 Temperature of winding
14.6 Mains transformer tested outside equipment
Construction Review:Construction review performed with satisfactory results.

Edition 11: Project 80204830
Update cCSAus report 1018431 (rev 80133090) for the addition of alternate controller to LoCC.
No testing was deemed necessary. The new alternate controller has similar ratings to the previously used alternate
controllers and is rated for a max temperature much greater than has been measured on previously used
controllers. In addition, the new alternate controller is certified as cULus.
Construction Review:Construction review performed with satisfactory results.
This Edition supersedes all previous editions.

QD-1399 Rev 2024-02-01© 202404:38:53
CSA Group.
rightsby:reserved.
Printed on: 24 Jul 2024,
pm; All
Printed
TVLAHOPOULOS.

This copy expires on 26 Jul 2024 at 04:38:25 pm.
CPD-ELTF-1008 Rev D
AS-Series CSA Safety Test Report and Certificate (non-medical products)
Model: AS Series Oxygen and Nitrogen Systems UL 61010 1/CAN/CSA C22 2 No 61010 1 12

Signature Manifest
Revision: D
Document Number: CPD-ELTF-1008
Title: AS-Series CSA Safety Test Report and Certificate (non-medical products) Model: AS Series Oxygen andNitrogen Systems, UL 61010-1/CAN/CSA-C22.2 No. 61010-1-12
Effective Date: 05 Jun 2024
All dates and times are in Eastern Standard Time.

CPD Nitrogen ConcentratorModel: AS-G-N2_CAN/CSA-C22.2 No. 61010-1-12 for CSA Mark_ Test Reportand Attachments

Collaboration
Name/Signature
Ted Vlahopoulos(TVLAHOPOULOS)

Title

Date

Meaning/Reason

Regulatory Specialist

10 May 2024, 02:41:04 PM

Complete

Title

Date

Meaning/Reason

Project Manager

05 Jun 2024, 09:42:48 AM

Approved

Title

Date

Meaning/Reason

Manufacturing Manager

13 May 2024, 10:56:02 AM

Approved

Title

Date

Meaning/Reason

Regulatory Specialist

13 May 2024, 11:13:06 AM

Approved

Title

Date

Meaning/Reason

Manufacturing Manager
Project Engineer
Regulatory Specialist

05 Jun 2024, 09:42:48 AM
05 Jun 2024, 09:42:48 AM
05 Jun 2024, 09:42:48 AM

Email Sent
Email Sent
Email Sent

Regulatory Specialist

05 Jun 2024, 09:42:48 AM

Email Sent

Project Manager

05 Jun 2024, 09:42:48 AM

Email Sent

Engineering
Name/Signature
Utkarsh Bharadwaj(UBHARADWAJ)

Production Quality
Name/Signature
Greg Webb (GWEBB)Regulatory
Name/Signature
Sean Switzer (RSWITZER)

Notification
Name/Signature
Greg Webb (GWEBB)
Rupal Sharma (RSHARMA)
Sean Switzer (RSWITZER)
Ted Vlahopoulos(TVLAHOPOULOS)
Utkarsh Bharadwaj(UBHARADWAJ)

Printed on: 24 Jul 2024, 04:38:53 pm; Printed by: TVLAHOPOULOS.