Test Report Issue Date:

45461937 R1.0
15 May 2024

APPENDIX E – PROBE CALIBRATION©2024 Celltech Labs Inc. This document is not to be reproduced in whole or in part without the expressed written permission of Celltech Labs Inc.

Page 59 of 61Test Report Issue Date:

45461937 R1.0
15 May 2024

APPENDIX F – DIPOLE CALIBRATION©2024 Celltech Labs Inc. This document is not to be reproduced in whole or in part without the expressed written permission of Celltech Labs Inc.

Page 60 of 61

NCL CALIBRATION LABORATORIES
Calibration File No: DC-1904Project Number: 5921
Client.: Celltech
Address: 21 – 364 Lougheed Road, Kelowna, BC V1X 7R8, Canada

C E R T I F I C A T E OF C A L I B R A T I O N
It is certified that the equipment identified below has been calibrated in theNCL CALIBRATION LABORATORIES by qualified personnel following recognizedprocedures and using transfer standards traceable to NRC/NIST.

Validation Dipole (Head)
Manufacturer: SPEAGPart number: D2450V2
Frequency: 2450 MHz
Serial No: 825

Calibrated: 27/04/2021
Released on: 05/05/2021
This Calibration Certificate is incomplete unless accompanied by the Calibration Results Summary

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1904

Conditions
Dipole SN 825 was a re-calibration.
Ambient Temperature of the Laboratory:
Temperature of the Tissue:

21 C +/- 0.5C
21 C +/- 0.5C

Primary Measurement Standards
Instrument
Signal Generator
Network Analyzer
Spectrum Analyzer

HP
Keysight
Keysight

Serial Number
83640B
3844A00689
E5063A
MY54502902
N9030B
MY57140772

Cal due date
Sept. 17, 2022
Mar. 9, 2023
Apr. 20, 2023

Attestation
The below named signatories have conducted the calibration and review of the data which ispresented in this calibration report.

Dipole SN: 825

Page 2 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1904

Calibration Results Summary
The following results relate the Calibrated Dipole and should be used as a quick reference
for the user.

Tissue Validation
TissueFrequency

Head2450 MHz

Dielectricconstant, r
40.73

Conductivity, [S/m]1.86

Return Loss-19.83 dBImpedance
43.26 

SWR:
1.23U

Electrical Specification
TissueHeadFrequency
2450 MHz

System Validation Results
TissueFrequency

1-Gram SAR

10-Gram SAR

Uncertainty

Head2450 MHz

52.719 W/kg

24.015 W/kg

19.8%

HeadDipole SN: 825

Page 3 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1904

Introduction
This Calibration Report has been produced in line with the SSI Dipole Calibration Procedure
SSI-TP-018-ALSAS. The results contained within this report are for Validation Dipole
SN 825. The calibration routine consisted of a three-step process. Step 1 was a mechanicalverification of the dipole to ensure that it meets the mechanical specifications. Step 2 was an
Electrical Calibration for the Validation Dipole, where the SWR, Impedance, and the Returnloss were assessed. Step 3 involved a System Validation using the ALSAS-10U, along withAPREL E-020 30 MHz to 6 GHz E-Field Probe Serial Number 225.

Referencesoooooo

IEEE Standard 1528:2013
IEEE Recommended Practice for Determining the Peak Spatial-Average SpecificAbsorption Rate (SAR) in the Human Head from Wireless Communications Devices:
Measurement Techniques
EN 62209-1:2016
Human Exposure to RF Fields from hand-held and body-mounted wirelesscommunication devices - Human models. instrumentation, and procedures - Part 1:
Procedure to measure the Specific Absorption Rate (SAR) for hand-held mobile
wireless devices
IEC 62209-2:2019
Human exposure to RF fields from hand-held and body-mounted wireless devices Human models, instrumentation, and procedures - Part 2: specific absorption rate(SAR) for wireless communication devices (30 MHz - 6 GHz)
D22-012-Tissue dielectric tissue calibration procedure
D28-002-Dipole procedure for validation of SAR system using a dipole
IEEE 1309 Standard for Calibration of Electromagnetic Field Sensors and Probes,
Excluding Antennas, from 9 kHz to 40 GHz

Conditions
Ambient Temperature of the Laboratory: 21 C +/- 0.5CTemperature of the Tissue:
21 C +/- 0.5CDipole Calibration uncertainty
The calibration uncertainty for the dipole is made up of various parameters presented below.
Tolerance, %2.00

MechanicalPositioning Error
Electrical

0.10
0.37

Tissue Permittivity
Tissue Conductivity

3.88
3.56Dipole Validation

1.70

Combined
k=2

4.81

Uncertainty,

Dipole SN: 825

Page 4 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1904

The Following Graphs are the results as displayed on the Vector Network Analyzer.
Electrical Calibration
Test

Head

S11 R/LImpedance
SWR-19.83 dB
43.26 1.23 U

S11 Parameter Return Loss
Head
Frequency Range 2326.59 MHz to 2450 MHz

Dipole SN: 825

Page 5 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1904

Smith Chart Dipole Impedance
HeadDipole SN: 825

Page 6 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1904

SWR
Head

Dipole SN: 825

Page 7 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL CALIBRATION LABORATORIES
Calibration File No: DC-1906
Project Number: 5921
Client.: Celltech
Address: 21 – 364 Lougheed Road, Kelowna, BC V1X 7R8, Canada

C E R T I F I C A T E OF C A L I B R A T I O N
It is certified that the equipment identified below has been calibrated in theNCL CALIBRATION LABORATORIES by qualified personnel following recognizedprocedures and using transfer standards traceable to NRC/NIST.

Validation Dipole (Head)
Manufacturer: SPEAGPart number: S5GHzV2Frequency: 450 MHz
Serial No: 1031

Calibrated: 27/04/2021
Released on: 05/05/2021
This Calibration Certificate is incomplete unless accompanied by the Calibration Results Summary

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

Conditions
Dipole SN 1031 was a re-calibration.
Ambient Temperature of the Laboratory:
Temperature of the Tissue:

21 C +/- 0.5C21 C +/- 0.5C

Primary Measurement Standards
Instrument
Signal Generator
Network Analyzer
Spectrum Analyzer

HP
Keysight
Keysight

Serial Number
83640B
3844A00689
E5063A
MY54502902
N9030B
MY57140772

Cal due date
Sept. 17, 2022
Mar. 9, 2023
Apr. 20, 2023

Attestation
The below named signatories have conducted the calibration and review of the data which ispresented in this calibration report.

Dipole SN: 1031

Page 2 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

Calibration Results Summary
The following results relate the Calibrated Dipole and should be used as a quick reference
for the user.

Tissue Validation
Tissue
Head
Head
Head

Dielectricconstant, r
37.68
36.72
36.33

Conductivity, [S/m]
4.75
5.18
5.52

Frequency,
MHz5250
5600
5750

Return Loss,dB

Impedance,Ω-20.613-16.481-16.629

51.07
67.46

41.92

Frequency
5250 MHz
5600 MHz
5750 MHz

1-Gram SAR
79.47 W/kg
77.14 W/kg
75.54 W/kg

10-Gram SAR
22.91 W/kg
22.15 W/kg
22.01 W/kg

Frequency,
MHz
5250
5600
5750

Electrical Specification
Tissue
Head
Head
Head

SWR
1.21 U

1.35 U
1.35 USAR Validation
Tissue
Head
Head
Head

Uncertainty
22.3%
24.8%
24.5%

Dipole SN: 1031

Page 3 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

SAR Validation Results
TissueHeadFrequency
5250 MHz

1-Gram SAR
79.47 W/kg

10-Gram SAR
22.91 W/kg

Uncertainty
22.31 %

TissueHeadFrequency
5600 MHz

1-Gram SAR
77.14 W/kg

10-Gram SAR
22.15 W/kg

Uncertainty
24.8%

Dipole SN: 1031

Page 4 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

TissueHeadFrequency
5750 MHz

DC-1906

1-Gram SAR
75.54 W/kg

10-Gram SAR
22.01 W/kg

Uncertainty
24.5%

Dipole SN: 1031

Page 5 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

Introduction
This Calibration Report has been produced in line with the SSI Dipole Calibration Procedure
SSI-TP-018-ALSAS. The results contained within this report are for Dipole 1031. The
calibration routine consisted of a three-step process. Step 1 was a mechanical verification ofthe dipole to ensure that it meets the mechanical specifications. Step 2 was an Electrical
Calibration for the Validation Dipole, where the SWR, Impedance, and the Return loss wereassessed. Step 3 involved a System Validation using the ALSAS-10U, along with APREL E020 30 MHz to 6 GHz E-Field Probe Serial Number 225.

Referencesoooooo

IEEE Standard 1528:2013
IEEE Recommended Practice for Determining the Peak Spatial-Average SpecificAbsorption Rate (SAR) in the Human Head from Wireless Communications Devices:
Measurement Techniques
EN 62209-1:2016
Human Exposure to RF Fields from hand-held and body-mounted wirelesscommunication devices - Human models. instrumentation, and procedures - Part 1:
Procedure to measure the Specific Absorption Rate (SAR) for hand-held mobile
wireless devices
IEC 62209-2:2019
Human exposure to RF fields from hand-held and body-mounted wireless devices Human models, instrumentation, and procedures - Part 2: specific absorption rate(SAR) for wireless communication devices (30 MHz - 6 GHz)
D22-012-Tissue dielectric tissue calibration procedure
D28-002-Dipole procedure for validation of SAR system using a dipole
IEEE 1309 Standard for Calibration of Electromagnetic Field Sensors and Probes,
Excluding Antennas, from 9 kHz to 40 GHz

Conditions
Ambient Temperature of the Laboratory: 21 C +/- 0.5CTemperature of the Tissue:
21 C +/- 0.5C

Dipole Calibration uncertainty
The calibration uncertainty for the dipole is made up of various parameters presented below.
Tolerance, %2.00

MechanicalPositioning Error
Electrical

0.10
0.37

Tissue Permittivity
Tissue Conductivity

3.43
4.68

Dipole Validation

1.70

Combined
k=2

5.21

Uncertainty,

Dipole SN: 1031

Page 6 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

The Following Graphs are the results as displayed on the Vector Network Analyzer.
Electrical Calibration
Test

5250 MHz

5600 MHz

5750 MHzS11 R/L, dB-20.613-16.481-16.629

Impedance, Ω

51.07

67.46

41.92

SWR

1.21 U

1.35

1.35

S11 Parameter Return Loss
Head

Dipole SN: 1031

Page 7 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

Smith Chart Dipole Impedance
HeadDipole SN: 1031

Page 8 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

NCL Calibration Laboratories
Division of APREL Laboratories

DC-1906

SWRHeadDipole SN: 1031

Page 9 of 7
This page has been reviewed for content and attested to on Page 2 of this document.

Test Report Issue Date:

45461937 R1.0
15 May 2024

APPENDIX G - PHANTOM©2024 Celltech Labs Inc. This document is not to be reproduced in whole or in part without the expressed written permission of Celltech Labs Inc.

Page 61 of 61