LiDAR Sensor for Unmanned Laser ScanningRIEGL miniVUX-1UAV®

• very compact & lightweight(1.55 kg / 3.4 lbs)
• 360° field-of-view• robust aluminum housing,ready to be mounted on multi-rotor,
rotary-wing, and fixed-wing UAVs
• makes use of RIEGL’s unique echo
signal digitization and online
waveform processing• multiple target capability –up to 5 target echoes per laser shot• scan speed up to 100 scans/sec• measurement rate up to100,000 measurements/sec• mechanical and electrical interfacefor IMU mounting• exceptionally well suited tomeasure in snowy and icy terrains• user-friendly, application- andinstallation-oriented solutions for
integration

The RIEGL miniVUX-1UAV is an extremely lightweight airborne laser
scanner, designed specifically for integration with UAS/UAV/RPAS.
The small and sophisticated design of the stable aluminum housing
offers various integration possibilities with platforms that offer restricted
space or payload capabilities. The 360° field of view allows
complete acquisition of the environment.
An easy-to-remove SD card for data storage, and/or the option for
streaming the scan data via LAN-TCP/IP interface, in combination withthe modest power consumption of the scanner, enable straight-forward
integration with most UAS/UAV/RPAS types.
The RIEGL miniVUX-1UAV makes use of RIEGL’s unique Waveform-LiDAR
technology, allowing echo digitization and online waveform processing.
Multi-target resolution is the basis for penetrating even dense foliage.
As a further special feature, the wavelength is optimized for the measurement of snowy and icy terrain.
In addition to the stand-alone version of the miniVUX-1UAV, RIEGL alsooffers fully-integrated system solutions.

Typical applications include• Agriculture & Forestry• Glacier and Snowfield Mapping• Archeology and Cultural
Heritage Documentation• Construction-Site Monitoring• Landslide Monitoring

visit our website
www.riegl.com

Unmanned Laser Scanning

Data SheetMaximum Measurement Range vs. Target Reflectance RIEGL miniVUX®-1UAVOperating Flight Altitude AGL given for the following conditions:
FOV of +/-45°, target size ≥ laser footprint, average ambient brightness

Example: miniVUX-1UAV at 100,000 pulses/second, speed = 4 m/s,
range to target = 100 m, resulting point density ~ 40 pts/m²

RIEGL miniVUX-SYS System Integration Options
Besides of the stand-alone miniVUX-1UAV LiDAR engine, RIEGL offers also system solutions, combining the miniVUX-1UAV withIMU/GNSS systems of different performance and of different form factors as well as optional RGB camera systems. Additionally,a special add-on to the miniVUX-SYS allows for straight forward integration with your multi-rotor UAV, e.g. a DJI Matrice M600 / M300 RTK.
Please contact sales@riegl.com to get more detailed information.

RIEGL miniVUX-1UAV
with APX-15 UAV1)RIEGL miniVUX-1UAV
with APX-20 UAV (IMU82/IMU90)1)

RIEGL Integration Kit 600 / 300• IMU/GNSS unit integrated with LiDAR engine• total weight approx. 2 kg
• interfaces for up to 2 cameras• suited for integration into fixed-wing UAVs

• higher-grade IMU/GNSS unit

• add-on to the miniVUX-SYS coming

partly integrated (IMU82) respectively fullyintegrated (IMU90) with LiDAR engine• total weight approx. 2.5 kg / 2.3 kg• interfaces for up to 2 cameras• suited for integration into all types of UAVs

with shock-absorbing mounting-kit,
power supply module and cabling• total weight approx. 0.7 kg / 0.35 kg(without sensor and camera)• suited for integration into multi-rotor UAVs1) See technical details in the corresponding Applanix datasheet

2

Copyright RIEGL Laser Measurement Systems GmbH © 2022– All rights reserved.

Data SheetDimensional Drawings RIEGL miniVUX®-1UAV Stand-Alone

all dimensions in mm

RIEGL miniVUX®-1UAV Camera Options
RIEGL miniVUX-1UAV LiDAR Sensorequipped with APX-15 UAV1)

with two Sony Alpha 6000 cameras(oblique mount)with nadir-looking camera
e.g. Alpha 6000 camera orSony A7R III or Sony A7R IV

RIEGL miniVUX-1UAV LiDAR Sensor
equipped with APX-20 UAV (e.g. IMU82)1)

with two Sony Alpha 6000 cameras(oblique mount)integration of other 3rd party
cameras possible 2)with nadir-looking camera
e.g. Sony Alpha 6000 camera orSony A7R III or Sony A7R IV
integration of other 3rd party
cameras possible 2)

1) See technical details in the corresponding Applanix data sheet.
2) Multispectral camera, hyperspectral camera, thermal imaging sensor – more information on request.

Data Sheet

Copyright RIEGL Laser Measurement Systems GmbH © 2022– All rights reserved.

3

Technical Data RIEGL miniVUX®-1UAVLaser Product Classification

Class 1 Laser Product according to IEC 60825-1:2014
The following clause applies for instruments delivered into the United States:
Complies with 21 CFR 1040.10 and 1040.11 except for conformance withIEC 60825-1 Ed.3., as described in Laser Notice No. 56, dated May 8, 2019.

Range Measurement Performance
Measuring Principle

time of flight measurement, echo signal digitization,online waveform processing

Laser Pulse Repetition Rate PRR 1)			
Max. Measuring Range 2)
natural targets  ≥ 20 %			natural targets  ≥ 60 % 				natural targets  ≥ 80 % 						 Operating Flight Altitude AGL 1) 3)
Typ.			natural targets  ≥ 20 %			natural targets  ≥ 60 % 				Max. Number of Targets per Pulse 4)			

100 kHz				
170 m
290 m
330 m				100 m (330 ft)
160 m (525 ft)
5		1) Rounded values.
2) Typical values for average conditions. Maximum range is specified for flat targets with size in excess of the laser beam diameter, perpendicular angle of incidence, and foratmospheric visibility of 23 km. In bright sunlight, the max. range is shorter than under overcast sky.
3) Flat terrain assumed, scan angle ±45° FOV4)	If more than one target is hit, the total laser transmitter power is split and, accordingly, the achieveable range is reduced.

Minimum Range
Accuracy 5) 7)
Precision 6) 7)
Laser Pulse Repetition Rate 1)
Max. Effective Measurement Rate 1)
Echo Signal Intensity
Laser Wavelength
Laser Beam Divergence 8)
Laser Beam Footprint

3m
15 mm
10 mm
100 kHz
100 000 meas./sec. (@ 100 kHz PRR & 360° FOV)
for each echo signal, high-resolution 16 bit intensity information is provided
near infrared
1.6 x 0.5 mrad
160 mm x 50 mm @ 100 m

5) Accuracy is the degree of conformity of a measured quantity
to its actual (true) value.
6) Precision, also called reproducibility or repeatability, is thedegree to which further measurements show the same result.

7) One sigma @ 50 m range under RIEGL test conditions.
8) Measured at 50% peak intensity, 1.6 mrad corresponds to an increase of 160 mm of beam diameterper 100 m distance.

Scanner Performance

Scanning Mechanism
Field of View (selectable)
Scan Speed (selectable)
Angular Step Width   (selectable)

rotating mirror
up to 360°
10 - 100 revolutions per second, equivalent to 10 - 100 scans/sec
0.036° ≤   ≤ 0.36°

Angle Measurement Resolution

0.001°

between consecutive laser shots

Interfaces

Configuration, Scan Data Output &Communication with External Devices
GNSS Interface 9)General IO & Control 10)

2 x LAN 10/100/1000 Mbit/sec
WLAN IEEE 802.11 a/b/g/n
Serial RS-232 interface for data string with GNSS-time information,
TTL input for 1PPS synchronization pulse.
2 x TTL input/output, 1 x Remote on/off

Camera Interface

2 x GNSS RS-232 Tx & PPS, Power, Trigger, Exposure

Memory Card Slotfor SDHC/SDXC memory card 32 GByte (can be upgraded to 64 GByte)

9) internally available (not available with standard interface box)10) 1x externally available with standard interface box

General Technical Data

Power Supply Input Voltage / Consumption
Main Dimensions (L x W x H) / Weightwith Cooling Fan
without Cooling Fan

Humidity
Protection ClassTemperature Range 11)

11 - 34 V DC / typ. 18 W @ 100 scans/sec
243 x 111 x 85 mm / approx. 1.6 kg
243 x 99 x 85 mm / approx. 1.55 kg
max. 80 % non condensing @ 31°C
IP64, dust and splash-proof-10°C up to +40°C (operation) / -20°C up to +50°C (storage)11) Continuous operation at ambient temperature of ≥ 30°C (≥ 86°F)
requires a minimum amount of air flow at approx. 3 m/s.
For applications where a 3 m/s air flow along the cooling fins cannot
be guaranteed, the cooling fan has to be used.

RIEGL Laser Measurement Systems GmbHHorn, Austria
Phone: +43 2982 4211 | www.riegl.com
RIEGL USA Inc.
Winter Garden, Florida, USA
Phone: +1 407 248 9927 | www.rieglusa.com

Copyright RIEGL Laser Measurement Systems GmbH © 2022 – All rights reserved.
Use of this data sheet other than for personal purposes requires RIEGL’s written consent.
This data sheet is compiled with care. However, errors cannot be fully excluded and alternations might be necessary.

RIEGL Japan Ltd. | www.riegl-japan.co.jp
RIEGL China Ltd. | www.riegl.cn
RIEGL Australia Pty Ltd. | www.riegl.com
RIEGL Canada Inc. | www.rieglcanada.com
RIEGL UK Ltd. | www.riegl.co.uk

www.riegl.com
Data Sheet, RIEGL miniVUX-1UAV, 2022-09-27