Qx

Advanced Rasterizer forSD-4K SDI, HDR/WCG, SD-HD IP
Generation, Analysis and MonitoringQx Series - Technology to power change
From the moment you first power up a Qx, you’ll appreciate the attention to detail in aplatform designed to meet the increasing demands of monitoring and testing in SDI/IP
hybrid environments. The Qx is equally at home in master control rooms, OB and linktrucks, production studios, technical QC, product development, engineering compliancetesting and operational system monitoring. Whether you are working in SD, HD or UHD,SDR or HDR, SDI or IP, conventional or remote production, Qx brings together all theuser-configurability and advanced tools required for full operational flexibility whentransitioning to your next generation workflows.
The Qx provides an accessible user interface and intuitive toolsets that help with rapid
fault diagnosis and reduce the need for staff training. The comprehensive feature set
supports SD/HD/3G/6G/12G-SDI, 10G IP interfaces, and SD/HD/UHD, IP SMPTE 2022-6,
SMPTE 2110-10/20/30/31/40 with ST 2022-7, PCAP, Dolby E Decode and AMWA NMOS,
easing system design and future-proofing your investment.

Analyzer/Generator - Simultaneous Operation
The Qx provides Generation and Analysis toolsets that can be used simultaneously
either manually or under REST API control, enabling automated closed-loop testing of awide range of standards and SD*/HD/3G/6G/12G-SDI formats for engineering regression
testing and manufacturing. With a full suite of SDR Rec. BT 709/2020, plus native andmapped Wide Colour Gamut (WCG) HDR patterns in HLG, PQ, S-Log3 and SR-Live
formats, you are equipped for flexible broadcast SDR and HDR operation.
Generator and Analyzer video format, colorimetry and transfer function can all be
configured independently. You have the full flexibility to simultaneously send out a UHD
Rec BT.2100 HDR pattern with up to 128 channels of audio, and analyze the downconverted, down-mapped HD SDR Rec 709 return at the same time.

Compliance - it’s all about Test and Measurement

Developing products or commissioning the latest equipment is more than just implementation. Equipment has to be tested
against the required standards for it to be considered fit for purpose.

In the 12G-SDI world, noise floors are required to be much lower to ensure that accurate
and meaningful measurements can be taken. Qx SDI generation and measurement
technology has been specifically adapted for 12G applications. With its unique classleading SDI-STRESS toolset, sophisticated RTE™ (Real-Time Eye) multi-rate physicallayer display, and automated SMPTE compliance measurements, the Qx offers a single
product solution for SDI compliance verification.
If you are working in SMPTE ST 2110, with ST 2059 Precision Time Protocol (PTP), a
core IP toolset, available in the Qx offers an operator all of the IP confidence statusmonitoring in an intuitive and accessible manner. The optional IP-MEAS test suite
provides a comprehensive set of tools for compliance verification and commissioning ofyour IP systems and equipment.
Hardware-based timestamping locked to PTP ensures accurate realtime deterministic
timing measurements of media flows and ST 2110-21 buffer models.

2

Qx* Upcoming Software ReleaseApplications
Outside Broadcast

NEP UK selected Qx rasterizers for two of its new OB trucks,
for use at major events and sporting fixtures. Hybrid SDI/IP
capability was a key selling point for NEP enabling them toaccommodate clients whether they are using conventional
SDI or have made the move to IP. The ease of use of the Qx
was also a major factor, making it quick and simple for both
NEP engineering staff and freelancers to learn and use.
Engineering and Technical Director, NEP UK, said, “We’ve
been very happy with the reliability of PHABRIX test andmeasurement equipment in the past, so it was an obvious fit
to look at the Qx for these new IP-capable vehicles.”Sports and Live Events

PHABRIX recently concluded nine months of successful HDR
technology trials with BT Sport in the run-up to the launch of
BT Sport Ultimate. The Qx is now deployed to monitor andanalyze SDR and HDR Wide Color Gamut (WCG) material on
the live system. PHABRIX supported BT Sport, providing its
Qx rasterizers and technical expertise, as they developed andrefined their live production workflow for the launch of their
new HDR, UHD and Dolby Atmos® supported proposition. On
the bench PHABRIX collaborated with BT Sport to analyze
and provide suggested settings for SDR to HDR convertersand ‘tone mappers’ used in the trucks and throughout the
network.

Manufacturing & Compliance Testing

Mellanox Rivermax® development and regression testingteams have been using the Qx to provide simultaneous analysis of the SMPTE ST 2110 Video, Audio and ANC DATA flows
from their Rivermax® video streaming library for media andentertainment, running on Mellanox ConectX-5 and newer,
Network Interface Cards,” said Nir Nitzani, senior director SW
development at Mellanox Technologies.
“The ability to install the Qx in the machine room and remotely access and control the realtime measurements from several
sites has been an ideal fit with our engineering development
workflow."

www.phabrix.com

3

PHABRIX Qx - 10GbE / 12G-SDI Rasterizer
The flexible architecture of the Qx offers in-field license upgrades for SDI-UHD/4K,
PCAP, Dolby E Decode, HDR, AV test signal generation as well as engineering grade
data view and ANC packet inspection tools. A factory fitted hardware option provides
RTE™ realtime SDI eye and jitter analysis with the further option of a highly advanced
SDI-STRESS toolset.

If your focus is on a classical SDI based HD facility and you would like to future-proof your operation for
realtime IP, then the Qx will address your needs. The Qx features SD/HD/3G over SDI as standard, with 10G
IP, UHD-SDI (6G/12G). Dolby E Decode, IP Measure, PCAP and HDR available as optional in-field license
upgrades. Support for UHD/4K formats in SDI, including some HD/2K extended mode formats, can be added
as an optional license (for the full list of UHD standards supported, please see page 24).
A factory fitted SDI Eye and Jitter hardware option and the unique SDI-STRESS toolset provide all the tools
for SDI physical layer analysis and compliance testing.
The Qx’s optional IP toolset supports payloads on native 10G SFP+ interfaces using generic IP SFPs,
giving full access to be able to monitor and measure the IP traffic to SMPTE standards. In ST 2110 you cansimultaneously analyze ST 2110-10/20/30/31/40 JT-NM tested† flows with Class C Audio (up to 80 channels
at 125 μs packet time) all under ST 2022-7 Seamless IP Protection Switching (SIPS) and AMWA NMOS IS-04
discovery and IS-05 device connection management.
Optional in-field upgrades for IP-MEAS and PCAP provide a comprehensive set of tools for ST 2110
compliance verification and commissioning of your IP systems.

Also available in the Qx Series...
PHABRIX QxL - 25GbE/12G-SDI Rasterizer

KeyStandard
Optional

For realtime UHD IP workflows on 25G networks with video payloads up to
21Gbps, the class leading QxL provides support for 2110 and 2022-6 on generic
10G/25G SFP28 interfaces. The QxL is 10G IP-enabled as standard, with support
for simultaneous generation and analysis of a JT-NM TR 1001- 1:2018, ST 2110-20(video), up to four 2110-30 (PCM) and 2110-31 (AES transport) audio and a 2110-40
ANC media flow, all with 2022-7 Seamless IP Protection Switching (SIPS) and AMWA NMOS IS-04 discovery and IS-05 device
connection management.
HDR, PCAP, Dolby E Decode and IP-MEAS are available as in-field license upgrades. Support for UHD/4K formats for both IP and
SDI, including some HD/2K extended mode formats, and UHD 2110 Extended Mode formats (YCbCr/RGB 444, 8- / 10- / 12-bit; 48
to 60 Hz), can also be added.as optional licenses. The SDI Eye and Jitter hardware option and the unique SDI-STRESS toolset
provide all the tools for SDI physical layer analysis and compliance testing.

QxP - 25GbE/12G-SDI Portable Rasterizer
Introducing the latest member of the Qx test and measurement family - the QxP,
the world’s first portable, 12G-SDI, 25G-ST 2110, combined waveform monitor,
generator and analyzer. This provides all of the functionality of the QxL in a handy,
lightweight, portable 3RU chassis with an integral 1920 x 1200 7 inch LCD Screenwith multi-touch. If you prefer buttons or mouse control then you are free to use
any combinations of controls.
You can run all QxL instruments on the integral screen with minimal retraining. Plugin an external HDMI monitor and you have the same experience as if you wereusing the QxP as a conventional Rasterizer.

4

Qx†JT-NM Tested - For more details on the JT-NM Tested Program (Spring 2020) and its test results, please see:
http://jt-nm.org/jt-nm_tested/* Upcoming Software ReleaseSimplicity - an interface that puts you in control
The Qx’s innovative app style interface is a radical change from traditional test and measurement systems. Intuitive mouse control
with context-driven dropdown menus hides the complexity of modern SDI and IP systems providing an uncluttered view of criticalinformation. Instruments can be resized, the system auto-presenting more information as the screen area permits.
The Qx offers a fully flexible user-defined instrument layout, displaying up to 16 instruments on a single 1920 x 1080 display.
Individual instruments can switch between 1/16th, 1/4 or full screen. With an output frame rate of 50, 50.94 or 60 Hz to match thevideo format, the GUI has adjustable brightness for controlled lighting environments.

Instrument Tabs
• Tabs along the top of awindow provide quick
access to different
functional groups within asingle instrument

Screen Capture• SFTP and Browser network access to event
logs, screenshots, user presets

Instrument Tooltips• The UI displays context-driven

tooltips providing additional
information when hovering overparameters

Presets• Save multiple display layouts as presets andupdate as required• Save bespoke layouts for different
operational tasks• Use to change rapidly between different

Instrument Navigation• Each instrument includes a pop-up submenu,
giving you access to the configuration
parameters of that instrument

Network Time Protocol• Configure the unit to use a date and timetransmitted by a target Network Time
Protocol (NTP) server or set the time anddate manually in the Time and Date dialog.

screen layouts eg. Audio, HDR or IP focus

Instrument Launch Menu

• Provides access to the instruments andother system menus available on the unit

• Each Instrument available in the menu islisted alongside a designated icon

Numeric Slider & Scroll Bar• Adjust numeric values by dragging orError Highlighting• Errors are displayed in red fontscrolling the slider button• Mouse over the numeric field and scroll forfine control• Connect to USB keyboard, click and enter
specific alpha numeric values

www.phabrix.com

5

Introducing our New Waveforms
PHABRIX is pleased to announce the development of class leading Waveform Monitoring for the Qx Rasterizer. Using atechnique patented by PHABRIX to efficiently deliver a high-resolution image processing pipeline with support for deep color
sources up to 12-bits, this instrumentation delivers all the fine detail required fro Camera Shading, Image Grading or critical QC ofboth SDR and HDR content.
A choice of Overlay, Stacked and Parade display modes are provided each with the option of multi-colored, highlighted, green
or monochrome traces. The flexibility to display YCbCr, RGB, YRGB, YGRB and individual components is retained along withconnected instrument cursor linked to Picture and Data view, and user markers linked to Vectorscope. Single Line Mode and H
and V magnification are available for detailed inspection.
Luminance Nits scales and operation user-controlled Nits markers are provided for SDR, HLG, PQ, S-Log3, SR-live HDR formats.
Both SMPTE-narrow and full range operation are supported along with matrices for 709, 2020 and DCI P3 over the wide range
of YCbCr:422, RGB:444, SDI, 2110, SD*/HD/2K/UHD/4K/EUHD formats for which PHABRIX is famous.

New Overlay Mode
(RGB or YCbCR)
Accurate Reproduction of
Image Fine Detail While
Simultanously Retaining
Bright Instrument
Highlights

Full-width Waveform
Display Picture Modeto Match Picture Width6

Qx

Overlay YCbCr (Color)

Stacked YRGB (Color)

Stacked YCbCr (Color)

YRGB Parade (Color)

YRGB Parade (Color Highlight)

YCbCr Parade (Color)Standard Toolset
As standard, the Qx offers a flexible user-defined instrument layout displaying up to 16 simultaneous windows, and the ability tochange between bespoke layouts rapidly for different operational tasks with user presets.
Picture view, waveform monitor, vectorscope, 32 channel audio metering, decoded audio channel status information, detection ofcommon Dolby formats, ANC status and payload, on screen display of OP47 and CEA-608 in 708 closed captions and Ancillary
Time Code (ATC), Loudness monitoring, and advanced control and logging with human readable event logs, remote operator GUI
access over noVNC and a full REST API are all provided as standard.

Picture Display• Cursors linked to Waveform and Data View
• Action, graphics and user-definable custom
safe areas• 1/16, 1/4 or full size display

Analyzer - Waveform• YCbCr, YGBR and GBR parade modes• Cursor linked to Picture and Data View
• Single line mode linked to Picture Cursor• Configurable H and V Graticules• User markers• Overlay, Stacked, Parade, Single line, H & V
Mag, Brightness, Gamma, Persistence andColor, Highlight, Green and Monochrome
controls• 12-bit processing

Audio Status• 32 channel indication of audio type andpresence, PCM, Dolby E, DD, DD+, ED2• Decoded channel status information for up
to 128 channels• Clear indication of useful audio parameters
including CRCC, PCM/data, sample
frequency, word length• Channel Status data view (Hex)

CEA-608 Closed Caption
Decode and Display

CEA-608 Sous-titres Codes
Decoder et afficher

01:34:27:02

01:34:27:02

Auxiliary Data Decode

Analyzer - Picture Copy

• Closed Captions OP47, CEA-608 in 708• Primary Closed Caption decode picture• Secondary closed caption decode: Monitor

window• ANC Timecode with OSD• Date, V-chip, AFD and Input name
• SCTE 104 indication and logging

608/708 closed captions in a second
language, or compare different screen safeareas
• Independently manage overlay elements
including: Closed Captions, Picture Safe
Areas, V-chip, AFD, SCTE 104, Image Center
Crosshairs

Analyzer - Vectorscope

Analyzer - Audio Meters

Rec. 2020 and HDR formats• Custom user markers linked to Waveform• Center on target or custom user markers• 0.5x to 4x Mag, center on chosen target• Single line mode linked to Picture Cursor• Tooltip display of Cb, Cr and Hue Angle• IQ axis on/off• 12-bit processing

monitoring a block of up to 16 channels at a time,
for a total of up to 32 channels of audio metering• 2110 audio group display across up to four flows• Ballistics: PPM-I, PPM-II, Vu, Vu-Fr, Fast
• Scales: dBFS, dBu -18, dBu -20, BBC, DIN45406,
NordicN9• Adjustable peak hold times: Off, 0.1 s to Inf• Audio pair correlation meters, numerical level• Detection of Dolby E, ED2, DD, DD+, DE line pos• Stereo/mono audio preview bus• 75% and 100% Targets for ITU-R Rec. 709,

• Open one or two audio meter windows, each

Loudness Monitoring

Analyzer - Ancillary Status

short term and integrated loudness• User control of integrated, momentary andshort term targets• User-adjustable true peak alarm threshold• Loudness logging stored automatically• Grid View – clear visual overview, present/• EBU R128 and ITU-R BS.1770• Indicators for true peak, range, momentary,• SMPTE ST 291 VANC/HANC ancillary data
presence/status window
absent/fault indication• List View – ANC present list with location andstatus information for Checksum, Parity, DBN• Link to ANC Inspector• Tooltip provides ST 291 ANC type overview

www.phabrix.com

7Data View Analyzer with ANC Inspector

The engineering grade Data View Analyzer and ANC Inspector tools provide easy, accessible visualization of the data on
an SDI interface and associated ANC packets. Deep SDI data inspection with full freedom to inspect Active Picture, VANC,
HANC and API controls to read back Active Picture Data under automation control is included. Also featured is ANC packet
decapsulation and error reporting for detailed analysis and debug of ANC payloads.

Analyzer - Data View

• Allows analysis of complex faults• Detailed view of data words in the SDI
stream with tooltip hint

• Navigate function for rapid access to arequired line, pixel or TRS word

• Color-coding to help identification• Cursor linked to Picture and Waveform

ANC Inspector• Ancillary data packet analyzer• Link from ANC Status window• User-defined DID/SDID windowed search• Trigger on error, single shot, continuous• ANC packet capture with Hex view
• ANC packet decode view

I/O and Reference Configuration

SDI Video Timing &System Reference• Measurement of the timing of inputs againstreference• Indication of reference status and stability• Indication of the relative co-timing of input
SDI channels• Graphical and numeric display

System IO• Shows the status of signal inputs andoutputs, external reference, cable length,
and connector details• SDI mode: Select BNC or SFP I/O, cable
type, loop through and generator copy
outputs• IP mode: Active IP SFP receive inputs andtransmit outputs are indicatedAnalyzer - Video Standard CRC Analysis (SDI)• Display of detected SMPTE S352 Payload
ID for each SDI Link and Subframe• Manual override of S352 ID• Selection of SMPTE video format• Indication of S352 errors

audio output or AES input• SDI Input to AES Output de-embedder forboth PCM and Dolby encoded audio• AES Input signals can be routed to other
AES outputs providing up to three copy
outputs

Stats - SDI In• Cable length indication• Indication of data rate and clock divisor• Reporting of active and total pixel and line

• Detect active picture changes and view the• Y and C payload ID

failures, the last failure time, total analysis
time and error rates
active picture CRC to observe any changes
in the expected active picture CRC value
SMPTE RP168 compliance checking

Qxinterfaces• Audio meter monitoring pair, generator• Check for CRC errors on Y, C and ANC• Reporting of the number of SDI input• SDI switch line CRC masking control, for8

AES IO Config• Four versatile bi-directional AES unbalanced

countsRemote Access
Various methods are provided to enable you to establish a remote connection with your Qx system, depending on yourrequirements.

noVNC• Browser remote access using noVNC

technology to deliver 16 simultaneous
scalable instruments over a remote network

LLDP• Identify port and device to which the Qx

interfaces are connected• Restrict information communicated over
LLDP for IT security purposes• Also available in ST 2110 and ST 2022-6
boot modes

REST API• Control the Qx remotely over a network via
a REST API• Integrated control, monitoring andautomated manufacturer testing

Mgmt Interface Config• Manual or Dynamic Addressing modes• mDNS and DNS• Select Default Gateway from Media orManagement interfaces

Event Logger• SDI Input standard/status• SDI physical layer timing and alignment jitter• Rest API requests• IP-Tx, IP-Rx, Flow and SFP records• Reference Locking• Audio input presence

USB File Manager• Copy presets, instrument logs,

screenshots and user TIFF images to andfrom USB memory stick• Delete selected files• Control access to REST API and VNC

Qx Remote Connectivity• File Transfer: FTP or Browser access toscreenshots and PCAPs, User Test Patterns(TIFF), log files• Remote Software Product Updates• HDMI: UI video (1080p), UI audio (2-ch), local
mouse• SDI: UI video (1080p), UI audio (2-ch), local
mouse• noVNC: UI video (1080p low frame rate),
remote mouse with screenshots• KVM: HDMI or DVI (1080p compressed),remote mouse with screenshots• ST 2110: UI (-20), Audio 2-ch (-30)• UI audio available as analog on D26 (rearpanel)• Machine Control via JSON API• Many KVM Options available - including

Long Distance Connectivity, Cloud-based
solutions, multiple accesswww.phabrix.com

9

Optional Toolsets
Fast, automated 12G-SDI physical layer analysis [PHQX01E-3G]
The Qx Physical Layer Toolset is a factory-fitted option for fast 12G/6G/3G/HD/SD*-SDI† physical layer commissioning, testingand development. Its RTE™ (Real-Time Eye) Technology instantly highlights any SMPTE compliance issues and its realtime SDI
jitter window provides simultaneous monitoring across five specified frequency bands, jitter histogram and video trigger options.
Built-in automation control allows testing to be performed faster, more reliably and at lower cost. Included in the option are afull range of SDI eye measurements including amplitude, DC offset, transition times, overshoot and health indication with both
amplitude and time histograms, as well as choice of color, heat-map overlays and infinite persistence display.

SDI EYE Analysis

SDI Jitter Analysis• RTE™ (Real-Time Eye) for testing SMPTE compliance with indication of DC offset• Automatic measurements of: DC level, amplitude, rise and fall time, rise/fall overshoot, visual• Realtime SMPTE jitter measurements down

rise time indication• Amplitude and time histograms• Single or multiple eyes with choice of color, heat-map overlay and infinite persistence• Timing and Alignment jitter thermometers• User-definable time measurement cursors

Standard Eye Analysis

Realtime update of all
measurements

to 10 Hz• 10 Hz, 100 Hz, 1 kHz, 10 kHz, 100 kHz filters• H, 2H, F, V Trigger• Persistence control none to infinite• +/− 0.25 to +/− 64 UI vertical scale
adjustment• Jitter amplitude histogram

Alignment Jitter
Thermometer UIUser-selectable
Trace Colors(including Heatmap)

User-control of Horizontal
Timebase (1 to 20 Eyes)
Display High-pass Eye &Jitter Bandwidth Filter

Min/Max Scale and
DC Offset
Amplitude Markers(80% and 20%)

Rise/fall Cross:
Indication of
Measured Transitions

Vertical Histogram
Amplitude
Measurement

Automatic Rise/
Fall Overshoot
Percentage
DC Offset
Transition
Histogram† Note: Optional UHD SDI formats require PHQXO-UHD

10

Qx

Transition Times &Difference Between
Transition Times

Optional Toolsets
SDI-STRESS Testing [PHQXO-SDI-STRESS]
The advanced SDI-STRESS option is available for stress testing and R&D evaluations of SDI interfaces up to 12G. A
comprehensive API is included for rapid automation testing. The option includes the ability, under automation control, to insert
SDI clock jitter from 10 Hz to 10 MHz (128 UI max) peak-to-peak, mute any of the SDI outputs, and control the SDI scrambler,
sync-bit insertion, pre-emphasis, rise time and driver amplitude. The SDI-STRESS Eye amplitude measurement provides both
Shorth Mean or Mode, with a histogram overlay and a user-defined window for the exploration of eye amplitude. PseudoRandom Binary Sequence (PRBS) generation and analysis of PRBS-7, -9, -15, -23, -31 allows for deterministic measurement oflink Bit Error Rates (BER).

Adv. Generator Tools

PRBS Analyzer

Pathological Detector• Control of jitter insertion frequency and• Indication of PRBS cumulative received data

• Generator status indication of rate at which

amplitude• SDI scrambler and sync bit Insertion on/off• SDI Bit Error (BER) insertion tool
• Control of SDI driver amplitude +/− 15%• Control of pre-emphasis, rise/fall timeand PRBS type
• Generation of PRBS-7, -9, -15, -23, -31• Reported cumulative errors• Calculated Bit Error Rate (BER)

the video pattern generator is creating SDI
pathological conditions• Indication of PLL and EQ pathological rates
per second• Detection on each active SDI link
• Realtime GPI outputs of pathological detectfor external equipment triggering

Advanced Eye Analysis(Additional features with SDI-STRESS option)User-adjustable
Measurement Window

Eye Time Cursor (Two)User-adjustable
Windowed
Histogram (blue)

Automatic Amplitude
Measure with Modeor Shorth Mean

Eye Time Cursor (One)Measurements in Blue
Taken from User-adjustable
Measurement Window

Eye Time Cursor
Readings & Time
Delta
Automatic Measurement ofTransition Times and Transition
Time Delta

www.phabrix.com

11

Optional Toolsets
Audio and Video Generation [PHQXO-GEN]
Simultaneously generate and analyze a comprehensive set of SDI formats with the audio and video generation option. Moving
test patterns with up to 32 channels of embedded audio per link or sub-field (up to 128 channels on 12G interfaces) is included.
The Generator toolset option provides not only the core full screen SDI Pathological stress patterns (Eq, PLL, Clk, CheckField),
but uniquely also allows you to define a percentage combination of the SDI pathological and conventional generator patterns up
to full frame. Importing TIFF files for checking of HDR/WCG graphics or display and evaluation with user-created test images isalso included.

SDI Video Generation

SDI Audio Generation

andTest Pattern details• BNC output, SFP output and sub-image/fullimage mapping information• Video Reference, output offset adjustment
and Jitter instertion (with optional SDI
STRESS Toolkit) details• Reporting of SDI-STRESS pathologicalinsertion statistics• Moving test patterns (bouncing box)
• Import/display TIFF images

to help with channel identification• Choice of fixed or ramp levels – to help
with channel identification• Custom config of number of active audiogroups and channels• Master gain control• ST 2022-6: 32 channel audio generation
can be replicated in all sub frames
providing a total of up to 128 channels• Confirms generated Video Standard• Choice of fixed tones or chromatic scale –Pathological Generation(SDI Only)• Conventional SDI pathological stresspatterns, Eq, PLL and CheckField• New proposed SMPTE combined

pathological stress pattern: Eq + PLL + Color
Bars + Clock• Define a percentage combination of SMPTE
or SDI pathological and conventional
patterns up to full frame

SDI Configuration Dialog• 12G/6G/3G/1.5G 4K/UHD and 2K/HD SDI
signal generation• Support for Single, Dual, Quad link SDI

formats. Square division, 2SI, Level A & B• 422, 444, 4224 and 4444, YCbCr and RGB
formats, 10/12 bit• Supports the detection and analysis of
SMPTE Full Range video standards

12

QxOptional Toolsets
Dolby® E Decoder and Metadata Analyzer [PHQXO-DOLBY]The Dolby E Decoder and Metadata Analyzer option provides a clear and accessible view of the Dolby E metadata present
in a selected Dolby E or ED2 audio stream. It also enables you to check the correct timing of Dolby E packets in the audio
signal in an SDI or ST 2022-6 broadcast chain. You can check whether the Dolby E is created correctly and transferredtransparently by network equipment such as routers, switchers, satellite links, etc. You can also choose to monitor the Dolby®
audio from any of the SDI/2022-6 embedded audio, 2110-30/31 or AES inputs. The decoded output and downmix can bemetered, monitored, Loudness measured, and routed to AES outputs.

Dolby Metadata Analyzer
Source Pair/Channel ofthe Dolby Bitstream
Dolby Begin/End Gain
Values for Each Source
SignalProgram-dependantMetadata for Dolby E
Program 1
Program-dependent
Metadata for Dolby E
Program 2• Displays the Dolby E metadata present in

the selected Dolby E or ED2 audio stream• Enables you to check the correct timingof embedded Dolby E and ED2 in SDI and
2022-6 payloads• Check that the Dolby E metadata has beencreated correctly for multiple programs
using the easy to read metadata display• You can choose to monitor the Dolby audio
from any of the SDI, 2022-6 or 2110 input
embedded audio pairs/channels or the
AES input• Dolby stream CRC error detection anddisplay

Dolby Detection in Audio MeteringGroup Pair Labels
Stereo Pair Selectors
Mono Channel Solo
Bus Selectors

Detect any Number of
Dolby E, ED2, DD and
DD+ Pairs
Numeric Values ofEach Meter (dBFS)• Displays 16 audio meters together withpeak level indicators and indication ofaudio pair correlation• Dolby E, Dolby D and Dolby D+ streams
are detected by the system with Dolby
stream presence indicated in blue
• For an SDI or 2022-6 signal carrying
embedded Dolby E audio, the Dolby E
timing line number is also displayed below
the detector, either as an absolute value or
relative to the Ideal line number specified
for that video standard

Correlation Indicator for
Each Audio Pair
Dolby Timing Line Number
Absolute or Relative to Ideal

Dolby Decoder Metering and Status
8 Channels
Automatically Identified
from Dolby
Program Metadata
2 Channels for 2.0
Stereo Downmix of
Selected Dolby Program
8 Channel Audio
Metering for DolbyE, & 2
Channels for Downmix• When the Dolby E decoder is selected as

the metering source (2110, SDI or 2022-6
mode), the view of the analyzer changes todisplay the 8 channels of decoded Dolby E
audio as well as the stereo 2.0 downmix• The meter channel identification is
automatically configured from the Dolby
program metadata• Display of Dolby E source, line positioning(SDI, 2022-6), dialogue level and downmix
program source

Dolby Decoder Paneland Status Informationwww.phabrix.com

13

Optional Toolsets
ST 2110 and ST 2022-6 Monitoring [PHQXO-IP-STND]

The optional core IP feature set provides an operator with all of the ST 2110 confidence status monitoring in an intuitive andaccessible manner.
The toolset supports simultaneous decapsulation of one video, up to four audio and one ANC Data flows. Supported SMPTE
protocols include ST 2059 (PTP), ST 2110-20 (Uncompressed Video), -30 (PCM Digital Audio), -31 (AES3 Transparent Transport)
and -40 (ANC Data). ST 2022-7 seamless protection (SIPS) with AMWA NMOS IS-04, IS-05 and PTP system resource, is provided
over two media network interfaces using industry standard optical ethernet SFPs. Audio handling conforms to ST 2110-30 Class Cwith support for 48 kHz streams from 1 to 10 channels at packet times of 1 ms and 1 to 80 channels at packet times of 125 μs.

SMPTE 2110

SMPTE 2110 & 2022-6

Also provided is an indication of the timing relationship of each of the eight ST 2022-7 flows to PTP with status information, as
well as a ST 2022-7 status tool that reports the health and relative timing skew of each ST 2022-7 pair all with hardware timestamping.

14

SFP IP Network

SFP Information

IP Receive

physical network connection• Indication of SFP vendor and laser
characteristics• RX and TX power for debug of fiberconnectivity

receiver and user selection of the requiredflows• Indication of Qx locked status, Protocol, Src
and Dst IP and Port Numbers, SSRC, Packet
Counts, Sequence, payload and CRC errors• Configuration of Multicast Destination IP
addresses and subsequent Multicast Joinrequests• Reporting of presence of SFPs, SFP MAC• SFP status information for monitoring the

ST 2110 PTP Info - 2 port

ST 2022-7 Status

mode (multicast, hybrid w/o negotiation)• Indication of lock status• Grandmaster information including leader ID
and time source• Indication of estimated frequency andphase lock offsets• Indication of one step or two step traffic• Two independent PTP followers

seamless protection• Warning of ST 2022-7 flow-pair mismatch• Warnings of errors on flows and errors onreconstructed output and error rates per
second• Relative measure of Path Differential offlows on SFPB (Blue Network) relative to
SFPA (Amber Network)• Class A, B,C, D markers

and IP addresses (flow source IP address),and interface status• Tx and Rx packet counters for indication of
traffic activity• User configuration of SFP IP Addresses,
Masks, Gateway and DNS addresses• Control of PTP domain and communication

Qx• Indication of the health of ST 2022-7• Reporting of the IP Flows available to the

IP Flow Latency• Indication of media latency• Indication of relative timing of audio and
ANC flows with respect to video• Indication of relationship of underlying
media to PTP• External analog reference timing withrespect to PTP

Optional Toolsets
AMWA NMOS

A suite of AMWA NMOS tools that provide flexibility when integrating with an NMOS controller and associated network topology.
Supported protocols: IS-04 v1.0, 1.1, 1.2, 1.3 IS-05 1.02, 1.1 and IS-09 PTP domain. Provision of both in-band and out-of-band control
topologies with manual, mDNS, DNS-SD and DHCP. Receivers can be independently configured as single or dual NMOS end
points. NMOS troubleshooting is aided by the simultaneous views of the state of the Receiver Master and RTP Enables, SDP, andthe IS-05 parameters. The receiver auto-detected video format and audio packet time and channel count are compared with thereceived SDP information for diagnosis of the format information supplied by the SDP record.

NMOS Receiver Status

NMOS Receiver SDP

NMOS Receiver IS-05

receiver Master Enable, RTP Enables and
SDP records for each media interface• Available in 1/16 view - toggles with the SDP
view• Display of the Master, RTP and SDP of all
Receiver flows

improved readability• Display adapts with NMOS Dual or Single
receiver configuration (Dual shown)• Individual tabs for the display of the IS-05• At a glance overview of the state of the• Display of the active receiver SDP record• User configurable color highlighting for• Display of the active receiver IS-05
parametersparameters for each receiver flows• Human readable tree view of the IS-05 JSONwith expand/collapse for rapid navigation• Display adapts with NMOS Dual or Single
receiver configuration (Dual shown)

NMOS Setup• Manual, mDNS or DNS-SD discovery of theRegistry with DHCP
• Status reporting of registration and DNS
domain• Independent configuration of sender andreceiver as single or dual NMOS endpoints• NMOS node Enable/Disable• IS-09 PTP Domain Enable/Disable

2110 Format Setup• At a glance comparison of auto-detected,
SDP and manual format settings• User-configurable video format parameters
for ST 2110-20 flows• User-configurable audio format parameters

for ST 2110-30/31 flows includes packet timeand channel count• Automatic detection of audio format,
channel count and packet time

www.phabrix.com

15

Optional Toolsets
10G PCAP Tool [PHQXO-IP-PCAP]

This Packet Capture (PCAP) tool provides a flexible range of options for your capture of the live IP traffic on either a single orboth Media interfaces while in ST 2110 Mode. The PCAP data is then saved to USB memory stick for offline analysis using thirdparty network analysis tools. The PCAP data on the USB stick can be accessed remotely via Web Browser.

• Full line-rate capture at 10 Gbps on a single interface, back-to-back packets• Capture data on one or both media interfaces simultaneously up to 20 Gbps
• User control of packet capture size - e.g. Full payload or headers only with user control of the
Packet Capture size (12-1518 Octets)• Manual Start-Stop, Auto Start-Stop at specified time, Capture Start Delay• User controls for auto stop: Number of Packets, File size, Duration• Saves to USB stick with the option of Browser File transfer off the unit
• 1 GB PCAP max. file size on Qx.

IP Network Traffic Measurement [PHQXO-IP-MEAS]

An advanced engineering suite of tools for ST 2110 analysis and debug offers theprovision of up to four simultaneous dual Packet Interval Timing measurement
windows per input for easy visualization of network congestion and sender packet
distribution with max, mean and min inter-packet arrival times. Also included isdetailed data reporting of flow packet, clock rates and PTP timing relationship, as
well as the measurements of the ST 2110-21 Network Compatibility model (CINST) and
Virtual Receiver Buffer Model (VRX). Advanced measurement of IP flow latency and
RTP clock timing relationships for debug of Audio, Video and ANC alignment, source
PTP and encapsulation are featured.

Inter-packet Timing• Stream health reporting using histogram to

Advanced Media Timing - Video• TIme of First Received Packet of a Frame (video timing)• Receiver Buffer Margin with respect to TROdefault• Sender RTP offset

show the distribution of inter-packet arrival
times• Simultaneous reporting of ST 2022-7
primary and secondary flow
• Packet counts (log or linear scales) mapped
against arrival times (μs)
• Easy diagnosis of congestion with max,mean and min inter-packet arrival times

ST 2110-21

Advanced Media Timing - Media Latency• Numerical display of Mean, Min and Max latency measurements• Measured RTP clocks/s, Timestamps/s and RTP clocks/timestamp interval• Numerical display of Video to Audio and ANC relative latency measurement

16

Qx• ST 2110-21 measurement of Network

Compatibility Model (CINST) and Virtual
Receiver Buffer Model (VRX)• User control of VRX buffer read-scheduletiming• User control of CINST buffer drain rate

Optional Toolsets
Packet Interval Profile Generator[PHQXO-IP-NGT] (Requires PHQXO-GEN)

A ST 2022-6 packet generation tool for evaluating the ability of a receiver to handle a ST 2022-6 flow with jitter. Simulate IP
video network packet jitter under a variety of network conditions by providing the ability to adjust the transmission distribution
profile. View the interval timing distribution of the packets being generated, the number of packets being generated eachsecond, against the deviation of each packet interval from the expected interval time.

IP Transmit (ST 2022-6)• Configuration of Transmission flow

addresses, port numbers and SSRC

• Inter-packet jitter onto outgoing flow
• Gaussian or uniform distribution• Flow control on/off

High Dynamic Range (HDR) visualization & analysis toolset [PHQXO-HDR]

The Qx’s comprehensive HDR toolset includes a signal generator, CIE chart, Luma false color highlighting or heat map,
waveform monitor and vectorscope. All the main live production SDR and HDR formats are supported: Standard Dynamic
Range (SDR) BT.709, BT.2020 as well as HDR BT.2100 HLG, PQ and Sony S-Log3 and SR Live. The Waveform provides a
Cd/m2 (nits) graticule along with BT.2048 diffuse white markers. The flexible user controlled HDR heatmap offers sevensimultaneous programmable color overlay bands with presets for HDR and SDR ranges, plus a user custom preset. The CIE
1931 xy display provides overlays for BT.709, BT.2020 and ST.2086 gamut (P3) to enhance the visualization and analysis ofyour HDR/WCG content.
HDR Generator
An extensive set of test patterns include BT.2111 HDR color bars for HLG, PQ and SR Live as well as a full set of SDR 709
patterns mapped via ‘display light’ to each of the four HDR formats for line checks, comparative monitor set-up and theevaluation of HDR to SDR converters.

False Color Highlighting

Analyzer - CIE Chart

HDR Waveform• Programmable Heat Map to highlight• CIE 1931 xy display• Single line mode linked to picture cursor• Pan and zoom
• ITU-R BT. 709, BT. 2020 and ST 2086 gamut• Waveform HDR graticules with Nits (Cd/m2)• BT. 2408 diffuse white markers• Support for BT. 1886, BT. 2100 HLG and PQ,

luminance zones providing quick
identification of shadows, skin or mid-tones
or specular highlights• Seven simultaneous programmable color
overlay bands• Presets for HDR and SDR ranges plus user
customSony S-Log3, SR Liveoverlays• Tooltip co-ordinate display• Support for BT. 1886, BT. 2100 HLG and PQ,
Sony S-Log3, SR Livewww.phabrix.com

17

QxP
Qx

Specifications

QxL

Formats supported (Generation, Analysis & Monitoring)

ST 2110-20/-30/-31/-40 / 2022-7 / 2022-6 Analysis over 10G Ethernet
ST 2110-20/-30/-31/-40 / 2022-7 / 2022-6 Analysis over 25G Ethernet
ST 2110-20/-30/-31/-40 / 2022-7 Generation
ST 2022-6 Generation
SDI IO
3G / 1.5G / 270M*-SDI HD / SD Analysis
3G / 1.5G / 270M*-SDI HD Generation
12G / 6G / 3G / 1.5G-SDI UHD Over SDI
25G IP Link Rates Over SFP28

Hardware and Software Options Supported
Audio / Video Generator (SDI, ST 2022-6, ST 2110)

RTE™ Real-Time Eye input (12G/6G/3G/HD/SD-SDI) x 1 (SDI input A) BNC
UHD / 4K Upgrade
SDI-STRESS Testing Toolset (Requires SDI Eye and Jitter Toolset)
Data View Analyzer with ANC Inspector
HDR/WCG Support
Dolby E Analysis
ST 2022-6, ST 2110-20/30/31/40 Decap with ST 2022-7 and Dual PTP
ST 2110 Network Traffic Measurement Toolset
ST 2022-6 Network Traffic Generator Toolset
ST 2110 Generator Toolset
PCAP
EUHD Formats over 25G 2110-20

SDI inputs / outputs

Factory Option
Factory Option††-(SDI, 2022-6, 2110)
Factory Option(SDI, 2022-6, 2110)
Factory Option(SDI)(SDI, 2110)(SDI, 2110)

Factory Option
Factory Option
Factory Option

Factory Option
Factory Option
Factory Option--

HDMI 2.0b
Micro BNCHDMI 2.0b
Micro BNC---

Ethernet inputs / outputs (accepts generic SFPs)

Audio inputs / outputs††(SDI, 2022-6)
Factory Option

4 x SDI inputs, SD / HD / 3G, 75 ohm terminated BNC
2 x SFP+ MSA / Non-MSA cages (12 Gbps copper or fiber SDI interfaces)
4 x SDI outputs, SD* / HD / 3G, 75 ohm BNC
2 x SFP+ 10G Cages (also MSA / Non-MSA 12Gbps copper or fiber SDI SFPs)
2 x SFP28 10 / 25G cages
2 x QSFP28 10 / 25 / 40 / 50 / 100G cages (For Future Functionality)

Factory Option
Factory Option-

4 x 75 ohm AES selectable I/O (26 pin high-density D-Type socket)
1 x Stereo analog audio output (26 pin high-density D-Type socket)
8 channel 48 kHz PCM audio on HDMI and SDI Instrument output

User interface

Integrated 1920 x 1200 7 inch LCD multitouch touchscreen
HDMI instrument output, 1920 x 1080, 4:4:4 RGB, Type A
SDI 3Gbit instrument out, 1920 x 1080, 4:2:2 YCbCr
ST 2110-20, ST 2110-30 instrument out, 1920 x 1080, 4:2:2 YCbCr
Remote Browser GUI access (noVNC)

HDMI 1.4
BNC-

Reference

2 x 75 ohm BNC looping reference input, tri-level or B&B with cross lock1 x 75 ohm Micro-BNC reference input, Tri/B&B with cross lockNetworking & control-

10/100/1000 BASE-T
8 x bi-directional GPI (26 pin high-density D-Type socket)

Monitoring

Internal Beeper
Integral Speaker / Headphone Socket-

Form factor

Size (Width x Height x Depth - excluding projections)
WeightElectricalPower consumption
4 Pin XLR power connector
AC power adapter
Integral PSU with IEC connector
Choice of External Battery V-mount or G-mount

Warranty

18--

211 x 44 x 253 mm
1.9 kg211 x 44 x 253 mm
1.9 kg215 x 132 x 330 mm
4.1 kg ‡50 W typical, 70 W max
12V nominal (10-18 V)
90-264 VAC, 120 W-

100 W typical, 120 W max
12 V nominal (10-18 V)
90-264 VAC, 120 W-

110 W typical, 160 W max
12 V nominal (11-18 V)
90-264 VAC, 160 W

Warranty (1 year)
Extended Warranty Package (3 - 5 years)

Qx* Upcoming Software Release† Requires SDI Factory Option‡ Excludes extenal battery

Standard

Optional

Qx Rear panel
Generator, Generator Copy or
SDI Input Loop Out
12G/6G/3G/HD/SD* BNC x4

Control
Network(10/100/1000
Base-T)

2x USBSDI Instrument
2.0
Output(to 1080p60)

Analyzer SDI Inputs
12G/6G/3G/HD/SD-SDI BNC x4

IP / SDI I/O(SFP+ MSA/NON-MSA 12Gbps
Copper, Fiber or 10G Ethernet)

12G/6G/3G/HD/SD-SDI Eye

Ref Loop
BNC x 2

HDMI Instrument
Output (to 1080p60)
Micro USB
(Factory Service)

Stereo Audio Out,LTC In, 8x GPI I/O,
4x AES I/O

Power
10 – 18 V

Qx Mezzanine Interface Card Block Diagram

www.phabrix.com

19Ordering Qx
Qx Chassis

Qx IP Options

PHQX01-3G

Qx 1U SD/HD/2K SDI Rasterizer, Analyser only

PHQXO-IP-STND

PHQX01E-3G

Qx 1U SD/HD/2K SDI Rasterizer with Eye & Jitter,
Analyser only

2022-6 Encap/Decap, 2110 Decap with 2022-7,
dual PTP, NMOS IS-04/05

PHQXO-IP-MEAS

PHQXM-01E

Qx Eye/Jitter return to factory upgrade (forPHQX01-3G chassis)

IP Measurement 2110-21, PIT histograms, timing(requires PHQXO-IP-STND)PHQXO-IP-PCAP

Qx SDI Options

PCAP 2x10Gbps line rate capture tool, 1GB max.
(requires PHQXO-IP-STND)PHQXO-IP-NGT

2022-6 IP Network traffic Generator Tool
(requires PHQXO-IP-STND, PHQXO-GEN)PHQXO-UHD

SDI 2K Extended + UHD/4K
4xHD/6G/4x3G/2x6G/12G-SDI

PHSFP-10GE-SR

SFP+ 10GBASE-SR Ethernet MM 850nm 300m
(requires PHQXO-IP-STND)PHQXO-SDI-STRESS

12G-SDI Stress Test Toolset (requires PHQX01E3G, PHQXO-UHD, PHQXO-GEN)

PHSFP-10GE-LR

SFP+ 10GBASE-LR Ethernet SM 1310nm 10km
(requires PHQXO-IP-STND)PHSFP-RT12-1310

SFP+ SM(10km) LC Non-MSA, Tx 1310nm, Rx 12601620nm 12G/6G/3G/HD/SD*-SDI

Qx SDI/IP Software Options
PHQXO-DOLBY

Dolby E Decoder, Metadata Analyser, LtRt/LoRo
downmix, metering

PHQXO-GEN

SDI/2022-6 AV Test Signal Generator(2022-6 requires PHQXO-IP-STND)PHQXO-HDR

HDR/WCG, CIE1931, HDR Heat map (PQ, HLG,
S-Log3, SR Live)Qx Fitting Kits / Cables
PHQXC-1

12G-SDI Eye Measurement Test Cable 1m

PHQXK119 inch rackmount kit (1x Qx/QxL chassis)

PHQXK219 inch rackmount kit (2x Qx/QxL chassis)

PHQXK39.5 inch rackmount kit (1x Qx/QxL chassis)

Qx Extended Warranty
PHQX01-3YEAR

PHQX01 Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHQX01-5YEAR

PHQX01 Upgrade from 1 to 5 Year Warranty(excludes SFP)

PHQX01E-3YEAR

PHQX01E Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHQX01E-5YEAR

PHQX01E Upgrade from 1 to 5 Year Warranty(excludes SFP)

* Upcoming Software Release

20

Qx

SDI SFP Interfaces
SDI SFP Interface

Link TypeSFP+B Link Rates

SFP+A Link Rates

SDI Transceivers Only
SFP Interface

N/A

N/A

Rx Ch1

Tx Ch1

SIngle Link: Rx/Tx

N/A

N/A

BNC A Rx
0.27/1.5/3/6/12

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: N/A

N/A

N/A

N/A

N/A

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

Rx Ch1

Tx Ch1Rx Ch1

Tx Ch1

Single Link: Rx/Tx

N/A

BNC C Tx
(Tx Copy) 0.27*/1.5/3/6/12

BNC A Rx
0.27/1.5/3/6/12

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: Rx/Tx

BNC C Rx
0.27*/1.5/3/6

BNC C Tx
0.27*/1.5/3/6

BNC A Rx
0.27/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

N/A

N/A

Rx Ch1Rx Ch2
N/A

One SDI Transceiver in Cage A

Two SDI Transceivers in Cages
A&B

SDI Dual Receivers Only

One SDI Dual Receiver in Cage A

Two SDI Dual Receivers in Cages
A&B

Single Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6/12

Dual Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6

BNC B Rx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

Rx Ch1Rx Ch2Rx Ch1Rx Ch2
N/A

Single Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6/12

Dual Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6

BNC B Rx
0.27*/1.5/3/6

Quad Link: Rx 1 2

BNC C Rx
0.27*/1.5/3

BNC D Rx
0.27*/1.5/3

BNC A Rx
0.27/1.5/3

BNC B Rx
0.27*/1.5/3

SFP Interface

N/A

N/A

Tx Ch2Tx Ch1
BNC A Tx
0.27*/1.5/3/6/12

SDI Dual Transmitters Only

One SDI Dual Transmitter in Cage A

Single Link: Tx

N/A

N/A

BNC B Tx
(Tx Copy) 0.27*/1.5/3/6

Dual Link: Tx

N/A

N/A

BNC B Tx
0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

Tx Ch2Tx Ch1

Tx Ch2Tx Ch1

Single Link: Tx

BNC D Tx
(Tx Copy) 0.27*/1.5/3/6

BNC C Tx
(Tx Copy) 0.27*/1.5/3/6/12

BNC B Tx
(Tx Copy) 0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: Tx

BNC D Tx
(Tx Copy) 0.27*/1.5/3/6

BNC C Tx
(Tx Copy) 0.27*/1.5/3/6

BNC B Tx
0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: Tx 2 3

BNC D Tx
0.27*/1.5/3

BNC C Tx
0.27*/1.5/3

BNC B Tx
0.27*/1.5/3

BNC A Tx
0.27*/1.5/3

SFP Interface

Rx Ch1Rx Ch2Tx Ch2Tx Ch1

Single Link: Rx/Tx

BNC C Rx
0.27*/1.5/3/6/12 4

N/A

BNC B Tx
(Tx Copy) 0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: Rx/Tx

BNC C Rx
0.27*/1.5/3/6

BNC D Rx
0.27*/1.5/3/6

BNC B Tx
0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

Two SDI Dual Transmitters in Cages
A&B

SDI Dual Transmitter plus SDI Dual Receiver

One SDI Dual Transmitter(Cage A) and One Dual SDI
Receiver (Cage B)

1 In quad link 2SI the Receivers will auto adapt to any order of sub-image to BNC mapping.
2 In quad link square division the sub image order is: BNC A:TL, BNC B:TR, BNC C:BL, BNC D:BR.
3 In quad link 2SI the sub image order is: BNC A:Sub 1, BNC B:Sub 2, BNC C:Sub 3, BNC D:Sub 4.
4 SD-SDI (270M (0.27G)) video is not supported for Mixed Dual Transmitters and Receivers* Upcoming Software Releasewww.phabrix.com

21

Supported 2K/HD/SD SDI Formats
The following SDI formats are standard on the Qx.
SMPTE Stnds.
Link (Content)

Interface

Resolution

Sampling
Structure

Pixel
Depth

Frame/Field Rate

HDR †

SDI

2022-6‡

ST 259 (ST 125)

SD (625i)

720 x 576

4:2:2 (YCbCr)

10

50i

-

A

OA

ST 259 (ST 125)

SD (525i)

720 x 485

4:2:2 (YCbCr)

1059.94i-AOAST 292 (ST 296)

HD

1280 x 720

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 30p, 29.97p, 25p,

O●

●

O●

O●

●

O●

ST 292 (ST 274)HD

720 x 485

4:2:2 (YCbCr)

1060i, 59.94i, 50i
30p, 29.97p, 25p, 24p, 23.98p

ST 292 (RP 211)

HD

1920 x 1080

4:2:2 (YCbCr)

10

30psF, 29.97PsF, 25psF, 24PsF, 23.98PsF

O●

●

O●

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p,
30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF

O●

●

O●

ST 292 (ST 2048-2)
ST 372 (ST 274)

HD

2048 x 1080

Dual Link HD

1920 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●

●

-

10

60i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

-

ST 372 (ST 274)

Dual Link HD

1920 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 372 (ST 274)

Dual Link HD

1920 x 1080

4:4:4 (YCbCr/RGB)

1260i, 59.94i, 50i
30p, 29.97p, 25p, 24p, 23.98p

O●

●

-

ST 372 (ST 274)

Dual Link HD

1920 x 1080

4:2:2 (YCbCr)

1260i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

-

ST 372 (ST 2048-2)

Dual Link HD

2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●

●

-

2048 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1030PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

-

ST 372 (ST 2048-2)
ST 372 (ST 2048-2)

Dual Link HD
Dual Link HD

2048 x 1080

4:4:4 (YCbCr/RGB)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

-

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

-

ST 372 (ST 2048-2)

Dual Link HD

2048 x 1080

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

ST 425-1 (ST 274)

3G Level A (1)1920 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●

●

O●

ST 425-1 (ST 2048-2)

3G Level A (1)2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●

●

O●

1280 x 720

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1060p, 59.94p, 50p, 30p, 29.97p

O●

●

O●

10

60i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

ST 425-1 (ST 296)

3G Level A (2)ST 425-1 (ST 274)

3G Level A (2)1920 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 425-1 (ST 2048-2)

3G Level A (2)2048 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1030PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

ST 425-1 (ST 274)

3G Level A (3)1920 x 1080

4:4:4 (YCbCr/RGB)

1260i, 59.94i, 50i, 30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

ST 425-1 (ST 2048-2)
ST 425-1 (ST 274)

3G Level A (3)3G Level A (4)2048 x 1080

4:4:4 (YCbCr/RGB)

1920 x 1080

4:2:2 (YCbCr)

1260i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

60p, 59.94p, 50p

O●

●

O●

ST 425-1 (ST 2048-2)

3G Level A (4)2048 x 1080

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

ST 425-1 (ST 274)

3G Level B-DL (I)1920 x 1080

4:2:2 (YCbCr)

10

ST 425-1 (ST 2048-2)

3G Level B-DL (I)2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●

●

O●

1920 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1060i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

12

60i, 59.94i, 50i, 30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

O●

●

O●

ST 425-1 (ST 274)

3G Level B-DL (II)ST 425-1 (ST 2048-2)

3G Level B-DL (II)2048 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 425-1 (ST 274)

3G Level B-DL (III)1920 x 1080

4:4:4 (YCbCr/RBG)

ST 425-1 (ST 2048-2)

3G Level B-DL (III)2048 x 1080

4:4:4 (YCbCr/RBG)

1230PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

ST 425-1 (ST 274)

3G Level B-DL (IV)1920 x 1080

4:2:2 (YCbCr)

1260i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

ST 425-1 (ST 2048-2)

3G Level B-DL (IV)2048 x 1080

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

1230PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

●

O●

† Note: Optional HDR formats require PHQXO-HDR‡ Note: ST 2022-6 formats require PHQXO-IP-STND
KEY● - Generator with PHQXO-GEN option and Analyzer
O - Optional
O● - Optional Generator with PHQXO-GEN option and Analyzer
A - Analyzer Only‘-’ - Not Available

22

Qx* Upcoming Software ReleaseSupported 2K/HD/SD IP Formats
The following 2K/HD ST 2110-20 formats are optional on the Qx with PHQXO-IP-STND
Resolution

Sampling Structure

Pixel
Depth

Frame/Field Rate

2110
HDR †

2110
SDR

720 x 576

4:2:2 (YCbCr)

10

50i

-

OA

720 x 485

4:2:2 (YCbCr)

1059.94i-

OA

1280 x 720

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1280 x 720

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1280 x 720

4:4:4(YCbCr/RGB)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1280 x 720

4:4:4(YCbCr/RGB)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:2:2(YCbCr)

860I, 59.94I, 50I

OA

OA

1920 x 1080

4:2:2 (YCbCr)

1060i, 59.94i, 50i

OA

OA

1920 x 1080

4:2:2(YCbCr)

1260I, 59.94I, 50I

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

8

60I, 59.94I, 50I

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

10

60I, 59.94I, 50I

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

12

60I, 59.94I, 50I

OA

OA

1920 x 1080

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

1920 x 1080

4:2:2 (YCbCr)

830PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

1920 x 1080

4:2:2 (YCbCr)

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF

OA

OA

1920 x 1080

4:2:2 (YCbCr)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

8

30PsF, 29.97PsF, 25PsF, 24psF, 23.97PsF

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

10

30psF, 29.97psF, 25psF, 24PsF, 23.97PsF

OA

OA

1920 x 1080

4:4:4(YCbCr/RGB)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

2048 x 1080

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

2048 x 1080

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

2048 x 1080

4:4:4(YCbCr/RGB)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

2048 x 1080

4:4:4(YCbCr/RGB)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

2048 x 1080

4:4:4(YCbCr/RGB)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

OA

2048 x 1080

4:2:2(YCbCr)

830PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

2048 x 1080

4:2:2(YCbCr)

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

2048 x 1080

4:2:2(YCbCr)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

2048 x 1080

4:4:4(YCbCr/RGB)

8

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

2048 x 1080

4:4:4(YCbCr/RGB)

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA

2048 x 1080

4:4:4(YCbCr/RGB)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

OA† Note: Optional HDR formats require PHQXO-HDR
KEY
O - Optional
A - Analyzer Only‘-’ - Not Available* Upcoming Software Releasewww.phabrix.com

23

Supported 4K/UHD Formats
The following SDI formats are optional on the Qx [PHQXO-UHD]
SMPTE Stnds.
Link (Content)

InterfaceSampling
Structure

Pixel
Depth

Frame/Field Rate

SDI
HDR †

SDI
SDR

ST 425-3 Annex B.1 (ST 2036-1)

Quad-link HD-SQ

3840 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-3 Annex B.1 (ST 2048-1)

Quad-link HD-SQ

4096 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-3 Annex B.2, (ST 2036-1)

Dual 3G-B-DS

3840 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-3 Annex B.2, (ST 2048-1)

Dual 3G-B-DS

4096 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-10 M1 (ST 2036-1)

6G-2SI

3840 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-10 M1 (ST 2048-1)

6G-2SI

4096 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (1) 2SI

3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (1) 2SI

4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (2) 2SI

3840 x 2160

4:4:4 (YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (2) 2SI

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (3) 2SI

3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (3) 2SI

4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (4) 2SI

3840 x 2160

4:2:2 (YCbCr)

1230p, 29.97p, 25p, 24p, 23.98p

O●

O●

12

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (4) 2SI

4096 x 2160

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (1) SQ

3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 425-5 Annex B (ST 2048-1)

Quad-link 3G-A, B (1) SQ

4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (2) SQ

3840 x 2160

4:4:4 (YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●

O●

10

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B, (ST 2048-1)

Quad-link 3G-A, B (2) SQ

4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (3) SQ

3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B, (ST 2048-1)

Quad-link 3G-A, B (3) SQ

4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (4) SQ

3840 x 2160

4:2:2 (YCbCr)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B (ST 2048-1)

Quad-link 3G-A, B (4) SQ

4096 x 2160

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (I)3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (I)4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (II)3840 x 2160

4:4:4 (YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●

O●

10

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (II)4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 2081-11 M1 ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (III)3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (III)4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1 ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (IV)3840 x 2160

4:2:2 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●4096 x 2160

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1 ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (IV)ST 2082-10 M1, ST 425-5 (ST 2036-1)

12G-2SI (I)3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 2082-10 M1, ST 425-5 (ST 2048-1)

12G-2SI (I)4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2036-1)

12G -2SI (II)3840 x 2160

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2048-1)

12G -2SI (II)4096 x 2160

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2036-1)

12G-2SI (III)3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2048-1)

12G-2SI (III)4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●

O●

12

30p, 29.97p, 25p, 24p, 23.98p

O●

O●

12

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2036-1)

12G-2SI (IV)3840 x 2160

ST 2082-10 M1 ST 425-5 (ST 2048-1)

12G-2SI (IV)4096 x 2160

† Note: Optional HDR formats require PHQXO-HDR

24

Resolution

Qx

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)
4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)
4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

KEY
O - Optional
O● - Optional Generator with PHQXO-GEN option and Analyzer‘-’ - Not Available

Supported 4K/UHD IP Formats [QxL/QxP Only]
The following 4K/UHD ST 2110-20 formats are optional and can be added to QxL/QxP only
[PHQXLO-UHD / PHQXPO-UHD].
QxL/QxP
Resolution

Sampling
Structure

Pixel
Depth

Frame/Field Rate

2110
HDR †

2110
SDR

3840 x 2160

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A

3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●3840 x 2160

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●3840 x 2160

4:4:4(YCbCr/RGB)

830p, 29.97p, 25p, 24p, 23.98p

OA

A

3840 x 2160

4:4:4(YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●●3840 x 2160

4:4:4(YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:2:2(YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p , 30p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.95p , 30p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:4:4(YCbCr/RGB)

830p, 29.97p, 25p, 24p, 23.98p

OA

A4096 x 2160

4:4:4(YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:4:4(YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●●

The following 4K/UHD ST 2110-20 extended formats are optional and can be added to QxL/QxP only
[PHQXLO-EUHD / PHQXPO-EUHD].
QxL/QxP
Resolution

Sampling
Structure

Pixel
Depth

Frame/Field Rate

2110
HDR †

2110
SDR

3840 x 2160

RGB:444860p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

RGB:4441060p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

RGB:44412660p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

YCbCr:444

860p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

YCbCR:444

1060p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

YCbCR:444

1260p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

RGB:444860p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

RGB:4441060p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

RGB:44412

60p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

YCbCr:444

860p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

YCbCR:444

1060p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

YCbCR:444

1260p, 59.94p, 50p, 48p, 47.97p

OA

OA

4K Formats† Note: Optional HDR formats require PHQXLO-HDR / PHQXPO-HDR
KEY● - Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
O - Optional
O● - Optional Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
A - Analyzer Only‘-’ - Not Available

www.phabrix.com

25

Dimensions and Installation
Desktop

Single Rack mount tray
with cover
PHQXK1

Dual Rack mountPHQXK2www.phabrix.com

E&O - As our policy is one of continuous improvement, we reserve
the right to vary details without prior notice.

This brochure is to be used for informational use only and is subject to change without notice and should not be construed as commitment
by Leader Electronics of Europe Limited. Leader Electronics of Europe Limited assumes no responsibility or liability for errors or
inaccuracies that may appear in this brochure.
Please visit www.phabrix.com for latest product information
July 2024