Centrox

PSA Oxygen Generator

The Centrox is an oxygen generator ideal for use in many applications. The Centrox’s
ability to deliver up to 95% oxygen concentration at flows of up to 32 SCFH (0.84
Nm3/hr or 15 LPM) make it a great fit for commercial uses requiring up to 50 psig
(345 kPa or 3.5 barg).

PSA Generator ModuleFeatures

Typical Applications● Self-contained (internal air compressors)● Brazing/Soldering● Microprocessor controlled● Environmental Remediation● Low operating cost

● Fish Farming● Automatic and unattended operation● Glass Work/Blowing● Wall or rack mount design● Ozone (Generator) Feed Gas● Heavy duty metal enclosures● Veterinary Medicine● Oxygen concentration monitor● Nano-bubble Feed Gas

Air Compressor
ModuleSpecifications
Product Flow1

32 SCFH (0.84 Nm3/hr or 15 LPM)

Product Pressure

50 psig (345 kPa or 3.5 barg)

Product Concentration

Up to 95%

Product Dew Point-100°F (-73°C)

Dimensions (W x D x H) (Nominal)● Air Compressor Module● PSA Generator Module

21 x 13 x 27 in (52 x 32 x 68 cm)
27 x 15 x 38 in (68 x 37 x 97 cm)

Weight● Air Compressor Module● PSA Generator Module

100 lb (45 kg)
160 lb (73 kg)

Sound Level

70 dBA

Physical Connection
Product Gas Outlet

B size oxygen demand valve

Ambient Operating Conditions

Locate the oxygen generator in a well-ventilated area that is protected from weather elements andremains between 40°F (4°C) and 112°F (44°C)

Control Power Requirements(Single Phase)

120 V ~ ±10%, 60 Hz, 11.0 A

60 Gallon Oxygen Receiver

1Dimensions (Dia. x H)

26 x 52 in (65 x 131 cm)

Weight159 lb (72 kg)

SCF (Standard cubic foot) gas measured at 1 atmosphere and 70°F (21ºC) / Nm3 (Normal cubic meter) gas measured at 1 atmosphere and 32°F (0°C) / SLPM (Standard liters per minute) gas measured at
1 atmosphere and 70°F (21°C)

60 Gallon Receiver

Note: All dimensions are nominal.

Ordering Information
Model
Centrox1
RequiredAccessories

Optional
Accessories

Part Number

Description

AS074-1

120 V ~ ±10%, 60 Hz with oxygen concentration monitor (LED indicator)

AS074-2

220 V ~ ±10%, 50 Hz with oxygen concentration monitor (LED indicator)

TA150-1

60 Gallon Oxygen Receiver

KI046-1

Accessory Kit (10 ft Stainless Braided Oxygen Hose & Primary/Secondary Oxygen Ball Valve Assembly)

KI470-1

Accessory Kit (10 ft Stainless Braided Oxygen Hose & Regulator Assembly with Flowmeter)

KI225-1

Accessory Starter Kit

TA212-1

Accessory Kit, Mini Oxygen Receiver

AN021-1

Oxygen Analyzer (Maxtec Handi)

AN005-1

Oxygen Analyzer (Maxtec Max O2 Plus)

AN075-1

Oxygen Analyzer/Sensor (Maxtec Max O2 Plus)

Shipping Information

Centrox

60 Gallon Oxygen ReceiverClass92.5

70

Commodity Classification
Number8421.39.8040

7311.00.0000

Dimensions (W x D x H)

34 x 52 x 50 in (86 x 132 x 127 cm) with Accessory Kits, Crated

28 x 30 x 56 in (71 x 76 x 142 cm)

Gross Weight559 lb (254 kg) with Accessory Kits, Crated

177 lb (80 kg)

Warranty: 1 Year Parts and Factory Labor****** An unprotected or inadequately ventilated environment, or improper control power may cause damage to the oxygen generator not covered under warranty.
1Specify oxygen flow and pressure at time of order.
All performance ratings based on an ambient temperature up to 100°F (38°C), up to 1,000 feet elevation, and 80% relative humidity.

© 2023 AirSep Corporation. All Rights Reserved.
AirSep reserves the right to discontinue its products, or change the prices, materials, equipment, quality,descriptions, specifications and/or processes to its products at any time without prior notice and with nofurther obligation or consequence. Any rights expressly stated herein are reserved by us, as applicable.

260 Creekside Drive
Buffalo, NY 14228-2075 U.S.A.
Tel: (716) 691-0202 ■ Fax: (716) 691-1255
www.airsep.com ■ info@airsep.com
ML-IND0008 F