M110

M-SERIES

M110

COMPACTUP TO 3.1 MW RF PEAK
POWER
MULTIPLE
DIAGNOSTICFEATURES
POWER FACTOR > 0.95
MINIMIZEDCONSUMABLESMAGNETRON MODULATOR
ScandiNova's M110 is a compact pulse Modulator
designed for system integration. The Modulator is
optimized for a wide range of different magnetrons
with an RF peak power ranging from 1.0 MW to 3.1 MW.
To facilitate system integration, the Modulator has asplit design with a Tank Unit and a Pulse Unit, and canbe placed in a very compact enclosure. The M110 isequipped with active power factor correction.
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers both a user-friendly interface and a capability forpulse shape modification. The pulse length, amplitude
and repetition rate can all be changed.
The M110 can be delivered as a pure high-power pulse
modulator or as part of an RF Unit including magnetronand related components. Different levels of service andsupport programmes are available.

THE M-SERIES
ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,
a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higher
efficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

M110

UNIT

VALUE

NOTES

Magnetron RF peak power

MW

1-3.1

Depends on choice of magnetron

Magnetron RF average power

kW

4.0

Maximum, depends on choice of magnetron

Modulator peak power

MW

6.2

Maximum

Modulator average power

kW

8Maximum

Pulse voltage

kV30-52

Pulse current

A30-120

RF pulse length, topμs

0.5-5

Depends on max average power

Pulse repetition rate

Hz0-500

Depends on max average power

Power factor-> 0.95

Water coolingl/min

8INTERFACECONNECTORSPECIFICATIONS

Mains power, three-phase

Terminal block (4-wire)

380-480±10 % VAC, 50/60 Hz

Mains power, single-phaseC20

115-230±10 % VAC, 50/60 Hz

Control interface

RJ45

Modbus TCP

Water cooling interface

Legris Push-in 12 mmTrig input

BNC

Electrical

Diagnostics

BNCPulse voltage & current signals

SIZE AND WEIGHT

UNIT

Weight, Pulse Unit

kg

Weight, Tank Unitkg

18-40 °C, non-condensing

NOTES

Other connector options available

VALUE3662

For more information, visit www.scandinovasystems.com/M110

The Standard Modulator Includes

Typical Magnetron Loads

Filament power supply

BVERI

GLVAC

Control system

VE2110

VE2110A

Control via Modbus TCP
Pulse sensors
Cable kit between Pulse Unit and Tank Unit
HV pulse cable

213 mmInternal trig generator
Factory acceptance test certificate
Manuals

459 mm399 mm

Pulse Unit

500 mmAdditional System Components
Circulator & RF loads
Directional coupler
Waveguide windows

266 mm

DOC-026570-01 / 2024-01-23

SPECIFICATIONSMagnetron

Nisshinbo

VE2098

M1603

GLM

M1466

GLM5193

MX7640

GLM5810

M1466N

GLM6090

M1466T

CPI

E2V

VMC3109

MG5193

VMC3136

MG6090MG6493
MG7095

Peaking components

Services
354 mm

430 mmTank Unit

Training in handling, operationand maintenance
Shipping
Installation and start-up

Contact ScandiNova for any requirements not covered by thisspecification.

Service contract

Information contained in this document is subject to change
without notice.

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power