Systematicpipetting
Sarpette® M2A perfect match
Compatible and compliant.
Robust and reliable.
Everything from one source.

Sarpette® M—Systematic Pipetting
SARSTEDT has been producing precision pipette tips formore than 50 years. Thanks to this long-standing experience,and technological expertise as well as true to the motto:“Everything from a single source”, we have developed thenew Sarpette® M, an innovative micro litre pipette that isperfectly tailored to today’s applications in the laboratory.
Sustainability and environmentally responsible behaviour are
gaining an increasing importance in these days and age.

Flexibly adjustable tip ejector

Precision fit coneSystematic pipetting  ///  SARSTEDT

3

Sarpette M®

For high-precision pipetting

Volumecolour coding
Ergonomic ejector push button

Precise andsmooth liquidintake

Volume display
on the front

Integralcalibration system

Try it out for free andwithout obligation!
sarpette.
sarstedt.com

4

Ergonomic design
Free moving and smooth running pipetting
push button. Human-engineered andconvenient ejection position. Flexible andadjustable tip ejection.
User-friendliness is a Must—particularly ergonomic design
In daily applications, in research, diagnostics and in any
laboratory, SARSTEDT precision pipettes stand out for
their excellent characteristics. Benefit from exceptional easeof handling, lightness and durability.

Systematic pipetting  ///  SARSTEDT

5

Unique position of the ejector

Ideal pipetting

The ejector directly above the hand rest enables a particularly
favourable physiological mode of operation. Thanks to thelarge ejector push button and the smooth-running pipetting
piston, the Sarpette® M guarantees particularly ergonomic
pipetting, its light weight an added benefit to prevent RSI(repetitive strain injury).

Thanks to the flexibly adjustable tip ejector, you can easilyadjust the ejector to the individual height of your pipette tips
with just one turn and little effort. This makes the Sarpette® M
compatible with a wide range of longer and shorter pipette tips
also from other manufacturers.

6

Reliable
Fast and accurate volume adjustment.
Accurate and precise pipetting.
Simple and safe calibration.

Fast and safe—easy volume
adjustment
To set the desired volume, simply turn thefree-moving and precise adjustment knob.
Audible click-stops and the freely rotatable thumbrest prevent unintended self-adjustment.

Systematic pipetting  ///  SARSTEDT

Precise and accurate
Exact pipetting results are a given fact with theSarpette® M. SARSTEDT pipette tips and yourSarpette® M provide a perfect match for all areasof application.

Calibration in a flash
Awkward calibration is a thing of the past. With thecalibration system of the SARSTEDT Sarpette® M,
you can now calibrate your Sarpette® M yourself—
quickly and safely. The precise system conforms
to all international standards.

78RobustStrong and durable construction.
Autoclavable and UV-resistant.
Convenient and quick cleaning.

No compromises in the choiceof materials
Quick and easy disassembly
Disassemble your Sarpette® M within seconds.
This allows you to clean and disinfect the pipette at
any time. Convenient and quick cleaning makes
contamination-free and safe pipetting even easier.

Systematic pipetting  ///  SARSTEDT

Fully autoclavable
The Sarpette® M can be easily autoclaved at 121°C.
The pipette is extremely resistant to chemicals andUV exposure.

Stable and durable construction
Thanks to the carefully matched materials andexcellent workmanship of the Sarpette® M, we canguarantee a warranty period of 3 years.

910

Ordering information
Sarpette® M performance values
VolumeInaccuracy (E%)

Imprecision (CV%)

Order no.
Volume range

Graduation

Min. vol.

Med. vol.

Max. vol.

Min. vol.

Med. vol.

Max. vol.

90.3100.002

0.1 – 2 µl

0.002 µl<+/– 6.0 %1<+/– 4.0 %<+/– 2.0 %< 5.0 %1

< 3.3 %

< 1.5 %

90.3100.010

0.5 –10 µl

0.01 µl<+/– 2.5 %2<+/– 1.8 %<+/– 1.0 %< 1.8 %2< 1.2 %

< 0.5 %

90.3100.020

2 – 20 µl

0.02 µl<+/– 2.5 %<+/– 1.8 %<+/– 1.0 %< 1.7 %

< 1.0 %

< 0.5 %

90.3100.100

10 – 100 µl

0.1 µl<+/– 1.5 %<+/– 1.2 %<+/– 0.8 %< 1.0 %

< 0.6 %

< 0.2 %

90.3100.200

20 – 200 µl

0.2 µl<+/– 1.5 %<+/– 1.1 %<+/– 0.6 %< 0.6 %

< 0.4 %

< 0.2 %

90.3100.000

100 – 1,000 µl

1 µl<+/– 1.5 %<+/– 1.0 %<+/– 0.5 %< 0.5 %< 0.4 %

< 0.2 %

90.3100.555

0.5 – 5 ml

0.01 µl<+/– 1.5 %<+/– 1.1 %<+/– 0.6 %< 0.6 %

< 0.5 %

< 0.3 %

90.3100.111

1 – 10 ml

0.01 µl<+/– 1.5 %<+/– 0.7 %<+/– 0.5 %< 0.5 %

< 0.3 %

< 0.2 %The performance values were determined using distilled water at a constant temperature (± 0.5 °C) between 20 and 25 °C in accordance with ISO 8655.
1) measured at 0.5 µl2) measured at 1 µl

Sarpette® M8 / M12 performance values
VolumeOrder no.

Inaccuracy (E%)

Volumerange

Graduation

Numberofchannels

Min. vol.

90.3108.010

0.5 –10 µl

0.01 µl890.3108.200

20 – 200 µl

0.2 µl90.3108.300

30 – 300 µl

90.3112.010

Med. vol.

Imprecision (CV%)
Max. vol.

Min. vol.

Med. vol.

Max. vol.

<+/– 3.5 %1 <+/– 2.5 %<+/– 1.5 %< 3.0 %1

< 2.0 %

< 1.0 %8<+/– 0.9 %1 <+/– 0.8 %<+/– 0.7 %< 0.6 %1

< 0.5 %

< 0.3 %

0.4 µl8<+/– 1.0 %1 <+/– 0.9 %<+/– 0.8 %< 0.6 %1

< 0.5 %

< 0.3 %

0.5 –10 µl

0.01 µl12<+/– 3.5 %1 <+/– 2.5 %<+/– 1.5 %< 3.0 %1

< 2.0 %

< 1.0 %

90.3112.200

20 – 200 µl

0.2 µl12<+/– 0.9 %1 <+/– 0.8 %<+/– 0.7 %< 0.6 %1

< 0.5 %

< 0.3 %

90.3112.300

30 – 300 µl

0.4 µl12<+/– 1.0 %1 <+/– 0.9 %<+/– 0.8 %< 0.6 %1

< 0.5 %

< 0.3 %The performance values were determined using distilled water at a constant temperature (± 0.5 °C) between 20 and 25 °C in accordance with ISO 8655.
1) measured at 10 % nominal volume

Systematic pipetting  ///  SARSTEDT

For more informationon the SARSTEDT pipette
tip range
www.refillrevolution.tips

Sarpette® M ordering information
Volume range [µl]Order no.

Packaging

Compatible SARSTEDT
pipette tips

0.1 – 2 µl

90.3100.002

1 pc./case70.3010.xxx
70.3020.xxx
70.3021.xxx

0.5 –10 µl

90.3100.010

1 pc./case70.3010.xxx
70.3020.xxx
70.3021.xxx

2 – 20 µl

90.3100.020

1 pc./case70.3030.xxx
70.3031.xxx

10 – 100 µl

90.3100.100

1 pc./case70.3030.xxx
70.3031.xxx

20 – 200 µl

90.3100.200

1 pc./case70.3030.xxx
70.3031.xxx

100 – 1,000 µl

90.3100.000

1 pc./case70.3050.xxx
70.3060.xxx

0.5 –5 ml

90.3100.555

1 pc./case70.1183.102
70.1183.002

1 – 10 ml

90.3100.111

1 pc./case70.1187.102
70.1187.002

0.5 –10 µl

90.3108.010

1 pc./case70.3010.xxx
70.3020.xxx
70.3021.xxx

20 – 200 µl

90.3108.200

1 pc./case70.3030.xxx
70.3031.xxx

30 – 300 µl

90.3108.300

1 pc./case70.3030.xxx
70.3031.xxx
70.3040.xxx

0.5 –10 µl

90.3112.010

1 pc./case70.3010.xxx
70.3020.xxx
70.3021.xxx

20 – 200 µl

90.3112.200

1 pc./case70.3030.xxx
70.3031.xxx

30 – 300 µl

90.3112.300

1 pc./case70.3030.xxx
70.3031.xxx
70.3040.xxx

Description

Order no.

Packaging

Pipette holder, rotable, for 6 SARSTEDT Sarpette® M single-channel pipettes
up to 1,000 µl

95.1760.200

1 pc. / inner box
4 pcs. / case

Pipette holder, for 3 SARSTEDT Sarpette® M single- and multi-channel pipettes

95.1760.201

1 pc. / inner box
4 pcs. / case

Description

Sarpette® M
single-channel

Sarpette® M
8-channel

Sarpette® M
12-channel

Pipette holders for Sarpette® M ordering information11

If you have any questions:We‘d be happy to help!

SARSTEDT AG & Co. KG
Sarstedtstraße 1D-51588 Nümbrecht
Phone: +49 2293 305 0
Fax:		 +49 2293 305 3992
export@sarstedt.com
www.sarstedt.com

BRO_20_898_0200_200_NF   This document may contain information on products that may not be available in particular countries   Technical modifications reserved

Visit our website:www.sarstedt.com