M100D

M-SERIES

M100D

DUAL-ENERGYCOMPACTUP TO 3.1 MW RF PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLES

THE M-SERIESMAGNETRON MODULATOR UP TO 3.1 MW
ScandiNova's M100D is a dual-energy pulse Modulator
in a very compact enclosure. Our dual-energy models
are for applications where there is a need to switch
between two different energy levels. The Modulator
is optimized for a wide range of different magnetrons
with an RF peak power ranging from 1.0 MW to 3.1 MW.
The M100D is also available as a single-energy (standalone) and integrated version (see M100, M100-i, andM100D-i).
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers a user-friendly interface. The pulse length,
amplitude and repetition rate can all be changed.
The M100D can be delivered as a pure high-power pulse
modulator or as part of an RF Unit including magnetronand related components. Different levels of service andsupport programmes are available.

ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higher
efficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

M100D

UNIT

VALUE

NOTES

Magnetron RF peak power, highMW

1-3.1

Depends on choice of magnetron

Magnetron RF peak power, lowMW

0.9-2.5

Depends on choice of magnetronMagnetron RF average power

kW

2.8Maximum, depends on choice of magnetron

Modulator peak power, highMW

6.2Maximum

Modulator peak power, lowMW

5.2Maximum

Modulator average power

kW8Maximum

Pulse voltage

kV30-52

Pulse current

A30-120

RF pulse length, topμs

0.5-5

Depends on max average power

Pulse repetition rate

Hz0-500

2x250 Hz, depends on max average power

Modulator Voltage Stability, RMS%

0.4

Verified on resistive load (see options)

Water coolingl/min818-40 °C, non-condensing

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

400±10 % VAC, 50/60 Hz

208/380/480 VAC

Mains power, single-phaseC20

230±10 % VAC, 50/60 Hz

115 VAC

Control interface

RJ45

Modbus TCP

Water cooling interface

12 mm

Legris Push-in

SwagelockTrig input

BNC

Electrical

OpticalDiagnostics

BNC

Pulse voltage & current signals

SIZE AND WEIGHT
Weight (incl. oil)

UNITkgVALUE
140

The Standard Modulator Includes

Services

Filament power supply

Training in handling, operationand maintenanceControl system

For more information, visit www.scandinovasystems.com/M100Control via Modbus TCP
Pulse sensors
HV pulse cable

605 mmInternal trig generator

647 mmFactory acceptance test certificate

12-32 mm

DOC-004506-01 / 2024-01-19

SPECIFICATIONS

401 mmManuals

Options
Pulse/RF diagnostics
Enhanced PRF range (500–1000 Hz)
Enhanced stability (down to 0.1 %)Digitizer

Contact ScandiNova for any requirements not covered bythis specification.
Information contained in this document is subject to change
without notice.

Peaking components

Additional System Components
Circulator & RF loads
Directional coupler
Magnet power supply
Waveguide windows

ShippingInstallation and start-up
Service contractz

Typical Magnetron Loads
BVERI

GLVAC

VE2110

VE2110A

Nisshinbo

VE2098

M1603

GLM

M1466

GLM5193

MX7640

GLM5810

M1466N

GLM6090

M1466T

CPI

E2V

VMC3109

MG5193

VMC3136

MG6090MG6493
MG7095Magnetron

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power