K100

With Canon S-band klystron E37323 or E37325

COMPACTFULLY EQUIPPEDUP TO 10 MW RF PEAK
POWER
FOR USE IN VARIOUS
RF APPLICATIONSINTEGRATED WITHCANON KLYSTRONTHE K-SERIES
ScandiNova's K-series
contains a range of klystronbased RF units with highreliability and performance,
a compact design and low
energy consumption.

TURN KEY RF UNIT 10 MW/10 KW, S-BAND
ScandiNova's K100 is a turn-key Radio Frequency (RF)Unit. By combining ScandiNova's solid-state pulsed
power technology with Canon's high quality klystrons,a very compact, high performance RF Unit has beencreated, providing up to 10 MW RF peak power. The
RF Unit is optimized for two different Canon S-band
klystrons: E37323 operating at RF frequency 2856 MHz,and E37325 operating at 2998.5 MHz.
The K100 is fully equipped with Klystron, Solenoid,
klystron accessories and supporting systems, suchas Solenoid Power Supply, Ion Pump Power Supply,
RF Amplifier, internal cooling system and radiation
shielding. All interlocks and essential diagnostics are
fully integrated, and the ScandiCAT™ control system
offers a safe and easy-to-use means of controlling theRF Unit.

The RF units utilize
ScandiNova's uniquesolid-state pulsed power
technology, integrated withdifferent types of klystrons,
usually operating in the RF
peak power range 3-100 MW.
The RF units have extended
diagnostics and features,
and are easy to install andoperate. They contain few
consumables and require
minimal maintenance.

There are several alternatives and options available
for the RF Unit, e.g. extended/premium performance,
different interfaces and RF components as well as
different levels of service and support programmes.
Excellence in pulsed powerwww.scandinovasystems.com

K100 Canon E37323/E37325

RF UNIT

UNIT

E37323

E37325EXTENDED

PREMIUM

Klystron RF peak power

MW0-100-10

Adjustable

Klystron RF average power

kW

0-10

0-10

Adjustable

RF frequency

MHz2856±52998±5

Set by RF source

RF pulse length, top

μs0-50-4

Adjustable

Pulse repetition rate

Hz0-200

0-250

0-500

0-1000Subject to max avg. and peak power

Voltage pulse to pulse stability

ppm< 100< 100< 50< 20

RMS 1000 consecutive pulses at < 200 Hz

RF pulse flatness%< ±1.5< ±1.5< ±1.0< ±0.75

Measured at max peak power

Radiation at 1 m distanceμSv/h< 10< 10<5<2Waveguide output value excluded

NOTES

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

360-430 VAC, 50/60 Hz, 55 kVA

200-220, 420-500 VAC, 50/60 Hz

Mains power, single-phase

Terminal block200-240 VAC, 50/60 Hz, 2.2 kVA

100-120, 50/60 Hz

Control interface

RJ45

ModBus TCP

Water cooling interface

Hose 1''80 dm /min, 20-30 °C

31-40 °C inlet water

Low conductivity water

Trig input

BNC

5-15 V into 50 Ω

HFBR: Optical/PW control

Pulse width can be set by trig pulse

Diagnostics

BNCPulse voltage & current signals

Waveguide output

WR-284

SF6RF amplifier input

SMA female

Nominal input power 0 dBm

SIZE, WEIGHT AND VOLUME

UNIT

VALUE

NOTES

100 Hz update rate

3Calibration according to test cert.
VacuumMax +15 dBm

The Standard RF Unit Includes

Options

Power Distribution Unit

Internal controllable RF source

Capacitive Charging Power Supply

Internal controllable phase shifter

Total system weightkg

1300- Modulator

kg

900

Solid-state Switch Unit

Neutron resistant Switch Unit

- Klystron, Solenoid and oil

kg

400

Pulse transformer and tank

Signal delay generator

Water cooling manifold andflow meters diagnostics

Additional radiation shielding

Total oil volumedm

208

Minimum lifting height of Klystronmm

appr. 2400

3

For more information, visit www.scandinovasystems.com/K100

Filament power supply
Local control panel(19” touch screen)

1263-1333 mm

2205 mm

Graphical user interface

Seismic reinforcement kit

Circulator & RF loads (SF6)RF Digitizer (FWD/RFL power/VSWR)

Waveguides

Directional coupler (SF6)

Waveguide window

Remote control via Modbus TCP

SF6 filling kit

Canon Klystron 10 MW

Spare part kit

Ion Pump Power Supply
RF Amplifier

Information contained in thisdocument is subject to change
without notice.

Reduced filament heating mode

Accessories

Canon Solenoid
705 mmKlystron conditioning tool

Digitizer for modulator diagnostics

Solenoid Power Supply
1659 mm

Front turned 90 º left

ScandiCAT™ control system

Factory acceptance test certificate
Manuals

1052 mm

K100 Canon E37323/E37325
DOC-023199-01 / 2024-01-23

TYPE OF KLYSTRONSYSTEM SPECIFICATIONS

Services
Training in handling, operation andmaintenanceFactory acceptance test participation
Site acceptance test
Shipping
Installation and start-up
Service contract

In collaboration withCanon Electron Tube Division
Klystron and Solenoid

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power