Clampex11 日本語マニュアル
株式会社インターメディカル

REV.240404
Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

2
目次
Clampex11 日本語マニュアル.................................................................................................................................................................................. 1
株式会社インターメディカル ....................................................................................................................................................................................... 1
1.

序説 ................................................................................................................................................................................................................ 12
1.1.

2.

新機能 ..................................................................................................................................................................................................... 13

1.1.1.

最大 50 個の Epoch を設定可能 ....................................................................................................................................................... 13

1.1.2.

Gap-Free モードでプリプログラミングを使用可能............................................................................................................................... 14

1.1.3.

4 つの Membrabe Test を同時にモニタリング可能 ............................................................................................................................ 15

1.2.

MiniDigi 1B .............................................................................................................................................................................................. 16

1.3.

pCLAMP ドキュメント ........................................................................................................................................................................... 17

1.4.

ユーティリティプログラム..................................................................................................................................................................... 18

1.5.

pCLAMP の歴史 ...................................................................................................................................................................................... 21

pCLAMP のセットアップ ................................................................................................................................................................................... 24
2.1.

インターフェース ................................................................................................................................................................................... 24

2.2.

OS 互換性 ............................................................................................................................................................................................... 25

2.3.

テレグラフの対応 ................................................................................................................................................................................... 26

2.4.

インストール........................................................................................................................................................................................... 27

2.5.

アップデート........................................................................................................................................................................................... 41

2.6.

アンインストール ................................................................................................................................................................................... 41

2.7.

ドングル .................................................................................................................................................................................................. 41

2.8.

ファイルの場所 ....................................................................................................................................................................................... 42

2.9.

セットアップの流れ................................................................................................................................................................................ 43

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

3
3.

4.

Digidata のセットアップ .................................................................................................................................................................................... 44
3.1.

Digidata1550, Digidata1550A, Digidata1550B のセットアップ............................................................................................................. 44

3.2.

Digidata1440A のセットアップ .............................................................................................................................................................. 45

3.3.

Digidata132xA のセットアップ .............................................................................................................................................................. 46

3.4.

Minidigi のセットアップ.......................................................................................................................................................................... 47

Axopatch 200B のセットアップ ......................................................................................................................................................................... 48
4.1.

Axopatch200B の設定 ............................................................................................................................................................................. 48

4.2.

Axopatch 200B の接続 ............................................................................................................................................................................ 49

4.3.

Telegraphed Instruments の設定 (Axopatch200B) ................................................................................................................................ 50

4.4.

Lab bench の設定 (Axopatch 200B)....................................................................................................................................................... 51

4.4.1.

Lab Bench - Signal リスト ................................................................................................................................................................. 52

4.4.2.

Lab Bench - Analog IN の設定 ......................................................................................................................................................... 53

4.4.3.

Lab Bench - Analog OUT の設定 ..................................................................................................................................................... 58

4.5.

Membrane Test Setup の設定(Axopatch 200B) ..................................................................................................................................... 60

4.6.

Protocol の Channel 設定 (Axopatch200B) ............................................................................................................................................ 61

4.6.1.

VC モードの設定 ............................................................................................................................................................................... 62

4.6.2.

IC モードの設定 ................................................................................................................................................................................ 65

4.7.
5.

External Command の設定 (Axopatch200B) ......................................................................................................................................... 68

Multiclamp 700B のセットアップ....................................................................................................................................................................... 69
5.1.

Mulriclamp700B の認識 .......................................................................................................................................................................... 69

5.2.

softpanel の認識 ...................................................................................................................................................................................... 70

5.3.

Multiclamp700B の接続........................................................................................................................................................................... 71

5.4.

Telegraphed Instruments の設定 (Multiclamp700B) .............................................................................................................................. 72

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

4
5.5.

5.5.1.

Lab Bench - Signal リスト ................................................................................................................................................................. 78

5.5.2.

Lab Bench - Analog IN の設定 ......................................................................................................................................................... 79

5.5.3.

Lab Bench - Analog OUT の設定 ..................................................................................................................................................... 83

5.6.

Membrane Test Setup の設定 (Multiclamp700B) .................................................................................................................................. 85

5.6.1.

Membrane Test Setup OUT 0 の設定 .............................................................................................................................................. 85

5.6.2.

Membrane Test Setup OUT 1 の設定 .............................................................................................................................................. 86

5.7.

Protocol の Channel 設定 (Multiclamp700B).......................................................................................................................................... 87

5.7.1.

VC モードの設定 ............................................................................................................................................................................... 88

5.7.2.

IC モードの設定 ................................................................................................................................................................................ 91

5.8.

6.

Lab Bench の設定 (Multiclamp 700B) .................................................................................................................................................... 77

External Command の設定 (Multiclamp700B) ....................................................................................................................................... 94

5.8.1.

VC モードの設定 ............................................................................................................................................................................... 94

5.8.2.

IC モードの設定 ................................................................................................................................................................................ 95

Axoclamp 900A のセットアップ ........................................................................................................................................................................ 96
6.1.

Axoclamp 900A の認識 ........................................................................................................................................................................... 96

6.2.

Softpanel の認識 ..................................................................................................................................................................................... 97

6.3.

Axoclamp900A の接続 ............................................................................................................................................................................ 98

6.4.

Telegraphed Instruments の設定 (Axoclamp900A)................................................................................................................................ 99

6.5.

Lab bench の設定 (Axoclamp 900A) .................................................................................................................................................... 102

6.5.1.

Lab Bench Signal リスト............................................................................................................................................................... 102

6.5.2.

Lab Bench - Analog IN の設定 ....................................................................................................................................................... 103

6.5.3.

Lab bench - Analog OUT の設定 ................................................................................................................................................... 106

6.6.

Protocol の Channel 設定 (Axoclamp900A) ......................................................................................................................................... 109

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

56.6.1.

TEVC モードの Inputs タブ .............................................................................................................................................................. 110

6.6.2.

dSEVC モードの Inputs タブ ............................................................................................................................................................ 111

6.6.3.

TEVC, dSEVC モードの Outputs タブ ............................................................................................................................................. 112

6.6.4.

IC, DCC, HVIC モードの Inputs タブ ............................................................................................................................................... 113

6.6.5.

IC, DCC, HVI モードの Outputs タブ ............................................................................................................................................... 114

6.7.

7.

6.7.1.

TEVC, dSEVC モードの External Command ................................................................................................................................. 115

6.7.2.

IC, DCC, HVIC モードの External Command ................................................................................................................................. 116

Axopatch 1D のセットアップ ........................................................................................................................................................................... 117
7.1.

Axopatch 1D の設定 .............................................................................................................................................................................. 117

7.2.

Axopatch 1D の接続 .............................................................................................................................................................................. 118

7.3.

Telegraphed Instruments の設定 (Axopatch 1D) ................................................................................................................................. 119

7.4.

Lab bench の設定 (Axopatch 1D) ........................................................................................................................................................ 120

7.4.1.

Lab Bench - Signal リスト ............................................................................................................................................................... 120

7.4.2.

Lab Bench - Analog IN の設定 ....................................................................................................................................................... 121

7.4.3.

Lab Bench - Analog OUT の設定 ................................................................................................................................................... 125

7.5.

Membrane Test Setup の設定(Axopatch 1D) ....................................................................................................................................... 127

7.6.

Protocol の Channel 設定 (Axopatch 1D)............................................................................................................................................. 128

7.6.1.

VC モードの設定 ............................................................................................................................................................................. 129

7.6.2.

IC モードの設定 .............................................................................................................................................................................. 132

7.7.
8.

External Command の設定 (Axoclamp900A)....................................................................................................................................... 115

External Command の設定 (Axopatch 1D) .......................................................................................................................................... 135

Axoclamp 2B のセットアップ .......................................................................................................................................................................... 136
8.1.

Axoclamp 2B の設定 ............................................................................................................................................................................. 136

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

6
8.2.

Axoclamp 2B の接続 ............................................................................................................................................................................. 137

8.3.

Lab bench の設定 (Axoclamp 2B) ........................................................................................................................................................ 138

8.3.1.

Lab Bench Signal リスト ................................................................................................................................................................. 138

8.3.2.

Lab Bench - Analog IN の設定 ....................................................................................................................................................... 139

8.3.3.

Lab bench - Analog OUT の設定 ................................................................................................................................................... 142

8.4.

Protocol の Channel 設定 (Axoclamp 2B) ............................................................................................................................................ 144

8.4.1.

TEVC モードの Channel 設定 ......................................................................................................................................................... 145

8.4.1.

SEVC モードの Channel 設定......................................................................................................................................................... 146

8.4.2.

BRIDGE, DCC モードの Channel 設定........................................................................................................................................... 147

8.5.
9.

External Command の設定 (Axoclamp2B)........................................................................................................................................... 148

OC-725C のセットアップ ................................................................................................................................................................................ 149
9.1.

OC-725C の接続 ................................................................................................................................................................................... 149

9.2.

Telegraphed Instruments の設定 (OC-725C) ....................................................................................................................................... 150

9.3.

Lab bench の設定 (OC-725C) .............................................................................................................................................................. 151

9.3.1.

Lab Bench Signal リスト............................................................................................................................................................... 151

9.3.2.

Lab Bench - Analog IN の設定 ....................................................................................................................................................... 152

9.3.3.

Lab Bench - Analog OUT の設定 ................................................................................................................................................... 154

9.4.

Membrane Test Setup の設定(OC-725C) ............................................................................................................................................ 155

9.5.

Protocol の Channel 設定 (OC-725C) .................................................................................................................................................. 156

10.

MEZ-7200 & CEZ-1200 のセットアップ ...................................................................................................................................................... 160

10.1.

MEZ-7200 & CEZ-1200 の接続 ......................................................................................................................................................... 160

10.2.

Lab bench の設定 (MEZ-7200 & CEZ-1200) ................................................................................................................................... 161

10.2.1.

Lab Bench Signal リスト ........................................................................................................................................................... 161

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

7
10.2.2.

Lab Bench - Analog IN の設定 ................................................................................................................................................... 163

10.2.3.

Analog IN #1 .............................................................................................................................................................................. 164

10.2.4.

Analog IN #2 .............................................................................................................................................................................. 165

10.2.5.

Lab Bench - Analog OUT の設定 ............................................................................................................................................... 166

10.2.6.

Analog OUT #1 .......................................................................................................................................................................... 167

10.3.

Membrane Test Setup の設定(MEZ-7200 & CEZ-1200) .................................................................................................................. 168

10.4.

Protocol の Channel 設定 (MEZ-7200 & CEZ-1200) ........................................................................................................................ 169

10.4.1.

ボルテージクランプの設定 ........................................................................................................................................................... 170

10.4.2.

カレントクランプの設定（電極抵抗測定用） ................................................................................................................................... 173

10.5.
11.

External Command の設定 (MEZ-7200 & CEZ-1200) ..................................................................................................................... 181

Protocol の設定（基本編） .......................................................................................................................................................................... 182

11.1.

Protocol の作成.................................................................................................................................................................................. 182

11.2.

Mode/Rate タブの設定 ...................................................................................................................................................................... 183

11.2.1.

Gap-free モード .............................................................................................................................................................................. 183

11.2.2.

Episodic stimulation モード............................................................................................................................................................. 186

11.3.

Input タブの設定 ............................................................................................................................................................................... 189

11.4.

Output タブの設定............................................................................................................................................................................. 190

11.5.

Trigger タブの設定 ............................................................................................................................................................................ 191

11.5.1.

Trigger タブの設定(Gapfree モード) ................................................................................................................................................ 191

11.5.2.

Trigger タブの設定(Episodic Stimulation モード) ............................................................................................................................ 192

11.6.

Waveform タブの設定 ....................................................................................................................................................................... 193

11.7.

Protocol の保存.................................................................................................................................................................................. 197

12.

細胞へアプローチする(Membrane Test) .................................................................................................................................................... 198

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

8
12.1.

Membrane Test Setup の設定 ........................................................................................................................................................... 199

12.2.

BATH 状態 ......................................................................................................................................................................................... 201

12.3.

PATCH 状態 ....................................................................................................................................................................................... 203

12.4.

CELL 状態 ......................................................................................................................................................................................... 205

13.

記録する .................................................................................................................................................................................................... 207

13.1.

保存先の設定 ..................................................................................................................................................................................... 207

13.2.

記録の開始 ........................................................................................................................................................................................ 208

14.

Protocol の設定（応用編） .......................................................................................................................................................................... 210

14.1.

ハムノイズを除去する（Digidata1550A, Digidata1550B のみ） ..................................................................................................... 210

14.2.

Epoch ごとにサンプリングレートを変更する（Gear Shift） .......................................................................................................... 214

14.3.

平均波形を記録する .......................................................................................................................................................................... 215

14.4.

さまざまな測定モード ...................................................................................................................................................................... 218

14.4.1.

Fixed-length events モード ........................................................................................................................................................ 218

14.4.2.

Variable-length events モード .................................................................................................................................................... 221

14.4.3.

High-speed oscilloscope モード ................................................................................................................................................. 224

14.5.

トリガー信号の応用 .......................................................................................................................................................................... 225

14.5.1.

Scope Trigger を使用する ........................................................................................................................................................... 225

14.5.2.

外部トリガーに同期してコメントを挿入する ................................................................................................................................... 226

14.5.3.

外部トリガーに同期して統計を計算する ....................................................................................................................................... 227

14.5.4.

外部トリガーに同期して記録を開始する ....................................................................................................................................... 228

14.5.5.

外部トリガーに同期して、Sweep, Run を開始する ....................................................................................................................... 229

14.6.

統計機能を使用する .......................................................................................................................................................................... 230

14.6.1.

Gapfree モード ............................................................................................................................................................................ 230

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

914.6.2.
14.7.

Stimulus, High-speed oscilloscope モード.................................................................................................................................. 231

コメント機能を使用する................................................................................................................................................................... 233

14.7.1.

ファイルにコメントを付ける ........................................................................................................................................................... 233

14.7.2.

測定中にコメントを挿入する ......................................................................................................................................................... 235

14.8.

入力信号を演算する .......................................................................................................................................................................... 238

14.9.

デジタル信号の設定（Digital Output）............................................................................................................................................. 241

14.9.1.

基本設定 ..................................................................................................................................................................................... 241

14.9.2.

交互にデジタル信号を出力する ................................................................................................................................................... 245

14.9.3.

連続したトレインを出力する ......................................................................................................................................................... 246

14.10.

14.10.1.

複数のステップパルスを作成する ................................................................................................................................................ 248

14.10.2.

時間にデルタを設定する .............................................................................................................................................................. 249

14.10.3.

ランプ波形を作成する .................................................................................................................................................................. 250

14.10.4.

トレイン波形を作成する ............................................................................................................................................................... 252

14.10.5.

記録データなどを刺激波形に使用する（Stimulus File） ................................................................................................................ 254

14.10.6.

ATF 形式で Stimulus File を作成する.......................................................................................................................................... 257

14.11.

15.

いろいろな波形を作成 ...................................................................................................................................................................... 248

オプション波形 ................................................................................................................................................................................. 265

14.11.1.

刺激波形の前にトレイン波形を追加する（Pre-Sweep Train） ....................................................................................................... 265

14.11.2.

リーク電流を除去する（P/N Leak Subtraction） ............................................................................................................................ 267

14.11.3.

Sweep ごとに時間やレベルを設定する（User List）...................................................................................................................... 270

14.11.4.

Sweep 間に Membrane Test を挿入する（Membrane Test Between Sweeps）.......................................................................... 279

エクスポート機能 ........................................................................................................................................................................................ 282
15.1.1.

clampex9 のプロトコルをエクスポートする ................................................................................................................................... 282

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

10

16.

15.1.2.

データを pCLAMP9（ABF1.8）にエクスポートする ........................................................................................................................ 283

15.1.3.

複数のデータを pCLAMP9（ABF1.8）にエクスポートする ............................................................................................................. 284

シーケンス機（Sequencing Keys） .............................................................................................................................................................. 288

16.1.

記録を繰り返す ................................................................................................................................................................................. 288

16.2.

異なるプロトコルを連結して記録する ............................................................................................................................................. 291

17.

ツール機能 ................................................................................................................................................................................................ 295

17.1.

Comment to Lab Book ...................................................................................................................................................................... 295

17.2.

Zero Stopwatch ................................................................................................................................................................................. 297

17.3.

Membrane Test を中断してトレイン波形を印加する ...................................................................................................................... 298

17.4.

Junction Potential .............................................................................................................................................................................. 301

17.4.1.

Junction Potential の使い方 ....................................................................................................................................................... 301

17.4.2.

液間電位を設定する。 ................................................................................................................................................................. 309

17.5.

18.

LTP Assistant ..................................................................................................................................................................................... 313

17.5.1.

LTP プロトコルの作成 .................................................................................................................................................................. 314

17.5.2.

Sequencing の設定 .................................................................................................................................................................... 315

17.5.3.

Input/Output チャンネルの設定 .................................................................................................................................................. 317

17.5.4.

Baseline プロトコルの設定 .......................................................................................................................................................... 318

17.5.5.

Conditionig プロトコルの設定 ...................................................................................................................................................... 321

17.5.6.

Sequencing key の設定.............................................................................................................................................................. 324

表示の設定 ................................................................................................................................................................................................ 331

18.1.

ツールバー、ステータバー、リアルタイムコントロールパネルの表示.......................................................................................... 331

18.2.

ウィンドウプロパティの設定 ........................................................................................................................................................... 333

18.2.1.

Scope ウィンドウプロパティ.......................................................................................................................................................... 334

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

11
18.2.2.

Online statistics ウィンドウプロパティ .......................................................................................................................................... 335

18.2.3.

Results ウィンドウプロパティ ........................................................................................................................................................ 336

18.2.4.

Lab Book ウィンドウプロパティ..................................................................................................................................................... 337

19.

ウィンドウの整列 ........................................................................................................................................................................................ 338

20.

お問い合わせ............................................................................................................................................................................................. 340

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

12
1.

序説
pCLAMP ソフトウェアは Molecular Devices の Axon Instruments Conventional Electrophysiology 製品であり、データ収集・分析の統合ソフトウェ

アです。各種の実験に対応できるようにデザインされ、電気生理学の実験と分析において標準となっています。pCLAMP 11 は最新のバージョンです。
pCLAMP ソフトウェアが提供する柔軟性は、電気生理学の伝統的なアプリケーション以外にも、多くのアプリケーションに適応することができます。
pCLAMP 11 は以下のように構成されています。
・Clampex：刺激波形の作成とデータ収集ソフトウェア
・Clampfit：データ分析ソフトウェア
・AxoScope：バックグランドでのチャートレコーディングソフトウェア
・Minidigi：2 チャネルのデジタイザ
Clampex ソフトウェアは、デジタル化されたデータを取得するための、多用途で強力なソフトウェアツールです。パッチクランプデータの収集に特化していますが、電圧またはクランプ反応を測定することに制限されていません。各チャンネルのスケールと単位を個別に変換できるので、どのような物質的パラメータでも測定することができます。例えば、シナプス電流、 光電子増倍管の蛍光シグナル、 ストレインゲージからの圧力、 その他のアナログ
シグナルをどのような組み合わせでも取得することができます。
Clampfit は、電気生理学のデータのための多種多様な統計、分析、変化、およびレイアウトツールを持つ強力なデータ分析ソフトウェアです。
AxoScope ソフトウェアと MiniDigi デジタイザを組み合わせて、別個のチャートレコーダとして使用することができます。例えば、Clampex と Digidata
を使用しながら、同時にバックグランドでチャートレコーダとして使用することができます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

13
1.1.

新機能

1.1.1. 最大 50 個の Epoch を設定可能
Epochを最大50までに設定可能で、より複雑な波形を設定することが可能です。
また、Epochの数値をコピー＆ペーストすることも可能で、より便利になりました。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

14
1.1.2. Gap-Free モードでプリプログラミングを使用可能
Gap-Freeモードでリアルタイムコントロールのボタンをクリックすると、プリプログラミングのダイアログが開きます。 各チャンネルのホールディン

グ電圧とホールディング時間を事前にプログラム可能です。また、デジタルビットをオンまたはオフに可能で、周辺機器にトリガー信号を出力して同期することができます。 最大50Epochまで設定可能で、 記録中に手動で値を変更することもできます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

15
1.1.3. 4 つの Membrabe Test を同時にモニタリング可能
4 つの Membrabe Test を同時にモニタリング可能で、マルチパッチクランプに対応しています。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

16
1.2.

MiniDigi 1B

MiniDigi 1B は低ノイズ、２チャンネルのデジタイザで、Axoscope ソフトウェアで使用するシンプルなデジタルチャートレコーダです。２つの独立した
16 ビットアナログ入力があり、各チャンネルのサンプリングは 1kHz です。PC と USB2.0 で接続し、電源は USB バス電源から供給されます。
フィルタリング
MiniDigi デジタイザは minmax またはアナログフィルタを使用することができます。minmax を選択すると最小と最大値の区間でデータが取り込まれます。アナログフィルタは、サンプリングレートの 1/5 ローパスアンチエイリアシングフィルタです。
項目

仕様

入力チャンネル数

2 single-ended

分解能

16-bit (1 in 65536)

取得レート

1 kHz

測定範囲

-10.000 V to +10.000 V

最大入力範囲

-50 V to +50 V

入力インピーダンス

1M

ゲイン

1

アンチエイリアスフィルタ

3 次ベッセル 1.5 kHz

消費電力<100mA

インターフェース

USB2.0OS

WinXp Pro 32bit , Win 7

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

17
1.3.

pCLAMP ドキュメント

pCLAMP には多くのドキュメントが提供されていて、プログラムを効果的に学ぶことができます。
ドキュメント内容

PDF Tutorial guides

ハードウェアとソフトウェアの場所
Start > All Programs > Molecular Devices > pCLAMP 11.1 > Clampex 11.1 Tutorial

セットアップガイド
PDF User Guide

入門、チュートリアル、詳解、Help > PDF Manual

アルゴリズム、参照
Online Help

概要、詳述、操作手順

Help > Clampex Help

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

18
1.4.

ユーティリティプログラム

pCLAMP は３つの主要プログラム（Axoscope, Clampex, Clampfit）に加えて、いくつかのユーティリティプログラムがあります。それらのプログラム
はすべて¥Program Files¥Molecular Devices¥pCLAMP 11.x¥にあります。Windows エクスプローラで pCLAMP 11.x フォルダを開き、それらのプログ
ラムをダブルクリックして起動できます。
Reset to Program Defaults (ClearRegistry.exe)
このプログラムは Start > Programs > Molecular Devices > pCLAMP 11.x > Reset to Program Defaults からでも起動できます。プログラムでエラーが発生したときに、この機能を使用します。また、Windows のレジストリを工場出荷時に戻したい時に使用します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

19
ABF Info (ABFInfo.hlp):
これはデータファイルのプロパティを確認するプログラムです。ABF データを選択して、簡易または詳細なプロパティを表示します。比較検討するためにファイルヘッダー情報をすぐに表示できますが、編集することはできません。Clampex か Clampfit でデータファイルを開き、File > Properties を選択することで表示することができます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

20
DongleFind (DongleFind.exe):
Axon Instruments のネットワークキーを含む、セキュリティーキー（ドングル）をチェックするプログラムです。このプログラムはセキュリティーキーを見つけ情報をレポートすることができます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

21
1.5.

pCLAMP の歴史
pCLAMP 6：パラレルポートドングル
プログラム

OS

機能

ClampexDOS

Episodic モードの取り込み

FetchexDOS

Gap free モードの取り込み

ClampfitDOS

ホールセルの解析

FetchanDOS

シングルチャネルの解析

pSTAT

DOS

シングルチャネルの解析

Clampex と Fetchex が Windows 版に統合
pCLAMP 7：パラレルポートドングル
プログラム

OS

機能

Axoscope

Windows

取り込み専用（Episodic モード不可）

Clampex

Windows

取り込み専用ClampfitDOS

ホールセルの解析

FetchanDOS

シングルチャネルの解析

pSTAT

DOS

シングルチャネルの解析

Clampfit が Windows 版に変更

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

22
pCLAMP 8：パラレルポートドングル
プログラム

OS

機能

Axoscope

Windows

取り込み専用（Episodic モード不可）

Clampex

Windows

取り込み専用Clampfit

Windows

ホールセルの解析

FetchanDOS

シングルチャネルの解析

pSTAT

DOS

シングルチャネルの解析
Featchan と pSTAT が Clampfit に統合
ドングルをパラレルポートと USB で選択可能、Minidigi-1A 付属

pCLAMP 9：パラレルポートドングルと USB ドングルを選択、Minidigi-1A 付属
プログラム

OS

機能

Axoscope

Windows

取り込み専用（Episodic モード不可）

Clampex

Windows

取り込み専用Clampfit

Windows

解析専用
USB ドングルのみ
Minidigi-1B 付属

pCLAMP 10：USB ドングル、Minidigi-1B 付属
プログラム

OS

機能

Axoscope

Windows

取り込み専用（Episodic モード不可）

Clampex

Windows

取り込み専用Clampfit

Windows

解析専用

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

23
pCLAMP 11：clampex USB ドングル, Clampfit ADV USB ドングル
プログラム

OS

機能

Axoscope

Windows

取り込み専用（Episodic モード不可）

Clampex

Windows

記録専用

Clampfit

Windows

解析専用

Clampfit ADVWindows

オプション解析

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

24
2.

pCLAMP のセットアップ

2.1.

インターフェース

各 Digidata には以下のインターフェースが必要になります。
Digidata

インターフェース

ボード

1200 A/B

ISA

付属の 1200A/B ボード

132xA

PCI

付属の Advansys ABP3925-00 SCSI ボード

1440A

USB 2

－

1550

USB 2

－

1550A

USB 2

－

1550B

USB 2

－

Mini-Digi 1A

USB 1

－

Mini-Digi 1B

USB 2

－

132xA は Adaptec 2940AU SCSI ボードや RATOC Systems U2SCXV SCSI-USB コンバータなどで動作できる場合がありますが、動作を保証するものではありません。
1550,1550,1550B は PC によって認識しない場合があります。対策として、USB2.0 ボードを増設します。Allegro USB PCIe (USB2-E)をご提供致しま
すので、ご連絡下さい。
認識できなかった PC：DELL Optiplex3020 win7pro, Thinkpad X240 win7pro
認識できた PC：Lets note CF-SX1 win7pro, HP Elite8300SF win7pro
認識できた増設ボード：Allegro USB PCIe (USB2-E)

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

25
2.2.

OS 互換性

pCLAMP 11 はバージョンによって対応 OS が異なります。、 Home Edition はサポートしていません。pCLAMP セットアッププログラムは自動的に
OS を認識して、OS に対応した pCLAMP をインストールします。
pCLAMP, Digidata, OS の対応表
Ver.

1200 A/B 132xA 1440A 1550 1550A 1550B 1A 1B

OS

6

○

×

×

×

×

×

×

×

MS-DOS 3.3 以上 (Win 不可)

7

○

×

×

×

×

×

×

×

95, 98, NT 4.0

8

○

○

×

×

×

×

×

×

95, 98, NT 4.0, 2000, XP Pro 32bit

9

○

○

×

×

×

×

○

×

98SE, NT 4.0, 2000, XP Pro 32bit

10.0-10.2

×

△

○

×

×

×

×

○

2000, XP Pro 32bit, Vista 32bit

10.3

×

△

○

×

×

×

×

○

2000, XP Pro 32bit, Vista 32bit, 7 Pro 32 & 64bit

10.4

×

×

○

○

×

×

△

○

2000, XP Pro 32bit, Vista 32bit, 7 Pro 32 & 64bit

10.5

×

×

○

○

○

×

△

○

2000, XP Pro 32bit, Vista 32bit, 7 Pro 32 & 64bit

10.6

×

×

○

○

○

○

△

○

XP Pro 32bit, Vista 32bit, 7 Pro 32 & 64bit

10.7

×

×

○

○

○

○

△

○

Win 7 Pro & Win 10 Pro, 32 & 64 bit

11.1-11.2

×

×

○

○

○

○

×

○

Win 7 Pro 32&64bit, Win 10 Pro 64 bit

11.3

×

×

○

○

○

○

×

○

Win 10 Pro 64 bit, Win 11Pro 64bit

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

26
132xA を Win7 で使用する場合の注意
10.0-10.3 で使用できますが、OS は 32bit のみ対応です。また、付属 SCSI ボードのドライバは下記リンクよりダウンロード可能です。
http://mdc.custhelp.com/app/answers/detail/a_id/20057
付属の SCSI ボード以外に、下記の SCSI ボードでも使用可能です。
http://www.tekram.com/dc-315u-ultra-scsi-host-adapter/

MiniDigi-1A は 32bit のみ対応です。
2.3.

テレグラフの対応

PC 制御増幅器(Multiclamp700 Series, Axoclamp900A)のテレグラフ機能は対応バージョンが異なります。
Multiclamp700A 8.2.0.159 以降
Multiclamp700B 9.0.2.018 以降
Axoclamp900A 10.0.0.65 以降

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

27
2.4.

インストールpCLAMP CD を読み込む前に、すべてのプログラムを終了して下さい。
インストール CD をドライブに挿入して、下図のようにインストーラを起動します。Next をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

28
インストール先を選択して、Next をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

29
ショートカットの作成先と使用できるユーザを選択して、Next をクリックして下さい。

pclamp11

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

30
コンピュータに Digidata が接続されていないことを確認し、Next をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

31
インストールが開始されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

32
以下の画面が表示される場合は、インストールをクリックします。

コマンドプロンプトが表示され、ドライバがインストールされます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

33
HASP HL Setup ウィンドウが表示され、インストール終了後に”HASP HL Setup”が開始されるとメッセージが表示されます。
OK をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

34
インストールが完了します。Finish をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

35
HASP Run-time Setup が開始されます。Next をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

36
Accept を選択し、Next をクリックします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

37
インストールの準備ができました。Next をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

38
インストールが開始されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

39
インストールが完了します。Finish をクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

40
デスクトップとプログラムにショートカットが作成されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

41
2.5.

アップデートMolecular Devices のホームページからマニュアルや最新版のアップデートをダウンロードすることができます。また、ダウンロードするにはアカウン
トを登録し、ログインが必要な場合があります。

2.6.

1.

アカウントを作成する。

http://mdc.custhelp.com/app/utils/create_account/2.

ログインする。

http://mdc.custhelp.com/app/utils/login_form/

3.

ダウンロードする。

http://mdc.custhelp.com/app/answers/detail/a_id/19559/

アンインストール

下記の手順でアンインストールを行います。

2.7.

1.

Start > Molecular Devices> pCLAMP 11.x を開きます。

2.

Uninstall pCLAMP 11.x Software を選択します。

ドングル

Clampex を使用するには下図（左）のような USB タイプの pCLAMP11 ドングルが必要になります。PC の USB 端子に接続して下さい。pCLAMP9 以
下では下図（右）のようなパラレルポートタイプのドングルもあります。

Clampex11

Clampex10, 9

Clampex9 old type

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

42
2.8.

ファイルの場所

バージョンによってファイルの場所が異なります。
pCLAMP 10.3 - 10.7 & pCLAMP11
関連ファイルファイルの種類

ファイルの場所

ユーザ

Data, Protocol¥Documents and Settings¥[user name]¥My Documents¥Molecular Devices¥pCLAMP¥…

システム

Lab Bench, System Lab Book

C:¥ProgramData¥Molecular Devices¥pCLAMP¥, User-defined telegraphs
アプリケーション

AxoScope, Clampex

C:¥Program Files¥Molecular Devices¥pCLAMP 10.x

, Clampfit, etc…

C:¥Program Files¥Molecular Devices¥pCLAMP 11.x

pCLAMP 10.0 - 10.2
関連ファイルファイルの種類

ファイルの場所

ユーザ

Data, Protocol¥Documents and Settings¥[user name]¥My Documents¥Molecular Devices¥pCLAMP¥…

システム

Lab Bench, System Lab BookC:¥Axon¥pCLAMP X, User-defined telegraphs
アプリケーション

AxoScope, Clampex

C:¥Axon¥pCLAMP X, Clampfit, etc…

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

43
2.9.

セットアップの流れ

以下の手順でセットアップを行います。2-8 はアンプによって設定が異なりますので、各アンプのセットアップの章をご参照下さい。
手順

項目

内容

1

Digidata のセットアップ

Clampex で Digidata を認識させます。

2

アンプの設定・認識

3

アンプと Digidata の接続

アンプと Digidata を接続します。

4

Telegraphed Instruments の設定

アンプの Gain 設定などを Digidata と通信する Telegraph 機能を設定します。

5

Lab Bench の設定

Digidata の入出力信号のスケーリングや単位を設定した Signal を作成します。

6

Membrane Test の設定

7

Protocol の Channel 設定

実験プロトコルを設定します。

8

External Command の設定

アンプの外部コマンドを設定します。

アンプを初期設定します。
Multiclamp700B などの PC 制御アンプは専用ソフトで認識させます。

Membrane Test とは記録前に、細胞にアプローチしながら、電極抵抗の測定や容量成分を補正するのに便利なオシロスコープ機能です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

44
3.

Digidata のセットアップ

3.1.

Digidata1550, Digidata1550A, Digidata1550B のセットアップ

1.

電源ケーブルを接続する。

2.

PC を USB2 ケーブルで接続する。（USB1 ケーブルは使用不可）

3.

Digidata1550、Digidata1550A の電源を入れる。

4.

Clampex11 を起動する。Digidata1550 は Clampex10.4 以上、Digidata1550A は 10.5 以上を起動する。

5.

Configure/ Digitzer を選択して Digitizer ダイアログを開きます。

6.

Change ボタンをクリックして Change Digitizer ダイアログを開く。

7.

Digitizer Type に Digidata1550 Series, Digidata1550A Serie, Digidata1550BSerie を選択して、
Scan ボタンをクリックする。

8.

Digidata バージョン、シリアル番号、ファームウェアバージョンが表示される。
Digidata1550A は HumSilencer Channel が表示される。

9.

Not present と表示される場合は、再度 Scan ボタンをクリックする。

10. OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

45
3.2.

Digidata1440A のセットアップ

1.

Digidata1440A に AC アダプタを接続する。

2.

Digidata1440 と PC を USB2 ケーブルで接続する。（USB1 ケーブルは使用不可）

3.

Digidfata1440A の電源を入れる。

4.

Clampex11 を起動する。（Windows 7 の場合は Clampex10.3 以上を起動する。）

5.

Configure/ Digitzer を選択して Digitizer ダイアログを開きます。

6.

Change ボタンをクリックして Change Digitizer ダイアログを開く。

7.

Digitizer Type に Digidata1440 Series を選択して、Scan ボタンをクリックする。

8.

Digidata バージョン、シリアル番号、ファームウェアバージョンが表示される。

9.

Not present と表示される場合は、再度 Scan ボタンをクリックする。

10. OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

46
3.3.

Digidata132xA のセットアップ

1.

PC に付属の Advansys ABP3925-00 SCSI ボードを取り付ける

2.

PC と Digidata132xA を SCSI ケーブルで接続する。

3.

Digidata132xA の電源を入れる。

4.

PC の電源を入れる。

5.

Clampex11 を起動する。

6.

Configure/ Digitzer を選択して Digitizer ダイアログを開きます。

7.

Change ボタンをクリックして Change Digitizer ダイアログを開く。

8.

Digitizer Type に Digidata132xA Series を選択して Scan ボタンをクリックする。

9.

Digidata バージョン、シリアル番号、ファームウェアバージョンが表示される。

10. Not present と表示される場合は、再度 Scan ボタンをクリックする。
11. OK ボタンをクリックして終了する。
12. キャリブレーションする場合は Configure をクリックして、
Configure Digidata 132x ダイアログを開きます。
13. Calibrate をクリックします。
14. キャリブレーションが終了したら、OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

47
3.4.

Minidigi のセットアップ

1.

PC と Minidigi を USB2.0 ケーブルで接続する。

2.

Axoscope10 を起動する。

3.

Configure/ Digitzer を選択して Digitizer ダイアログを開きます。

4.

Change ボタンをクリックして Change Digitizer ダイアログを開く。

5.

Digitizer Type に Minidigi 1B を選択して Scan ボタンをクリックする。

6.

Digidata バージョン、シリアル番号、ファームウェアバージョンが表示される。

7.

Not present と表示される場合は、再度 Scan ボタンをクリックする。

8.

OK ボタンをクリックして終了する。

9.

フィルタを選択する場合は Configure をクリックして、Configure Minidigi ダイアログを開きます。

10. フィルタを選択して、OK をクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

48
4.

Axopatch 200B のセットアップ

4.1.

Axopatch200B の設定

1.

Axopatch 200B にヘッドステージ、電源ケーブルを接続し、電源を入れる。

2.

Axopatch 200B を下表のように初期設定します。
設定項目

設定値

PIPETTE OFFSET

5.0

ZAP

0.5 msPIPETTE CAPACITANCE COMP.

最小（反時計回り）

SERIES RESISTANCE COMP. % PREDICTION

0 %, OFFSERIES RESISTANCE COMP. % CORRECTION

0 %, OFFSERIES RESISTANCE COMP. LAG

1 µs

WHOLE CELL CAP.

0 pF, OFFSERIES RESISTANCE

0 MΩ

HOLDING COMMAND

0 mV, x1, OFF

EXT COMMAND

OFF

METER

I

MODE

V-CLAMPCONFIG.

WHOLE CELL (β = 1)

OUTPUT GAIN:

α =1

LOWPASS BESSEL FILTER

5 kHz

LEAK SUBTRACTION∞ MΩ, OFF

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

49
4.2.

Axopatch 200B の接続

下表は Digidata との標準的な接続です。表に従って BNC ケーブルで接続して下さい。

Digidata の端子

Axopatch 200B の端子

ANALOG IN 0

SCALED OUTPUTANALOG IN 1

10 Vm

ANALOG IN 2

I OUTPUT

カレントクランプ時、膜電流の取り込み

ANALOG IN 3

DATA NOT VALID OUTPUT

Patch モードのリセット信号の取り込み

EXT. COMMAND INPUT

ボルテージクランプ時、コマンド電位信号の出力

ANALOG OUT 0

FRONT SWITCH

用途
ボルテージクランプ時、膜電流の取り込み
カレントクランプ時、膜電位の取り込み✔

ボルテージクランプ時、膜電位の取り込み

カレントクランプ時、コマンド電流信号の出力

TELEGRAPH INPUT 0

GAIN

Gain 設定を取得

TELEGRAPH INPUT 1

FREQUENCY

Filter 設定を取得

TELEGRAPH INPUT 2

CELL CAPACITANCE

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

必須

CELL CAPACITANCE 設定を取得

✔
✔

50
4.3.

Telegraphed Instruments の設定 (Axopatch200B)

Telegraph とはアンプと Clampex の通信を行う機能です。Clampex はアンプから Gain などの情報を受け取って、自動的に反映させます。
Configure/ Telegraph Instruments を選択して Telegraphed Instruments ダイアログを開きます。下図のように設定します。

①Digidata の入力チャンネルを②アンプを選択します。

選択します。

Axopatch 200B

Analog IN #0③Digidata の Telegraph チャンネルを選択します。
Gain → Telegraph Inputs 0Frequency→ Telegraph Inputs 1
Cm → Telegraph Inputs 2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

51
4.4.

Lab bench の設定 (Axopatch 200B)
Configure/ Lab Bench を選択して Lab Bench ダイアログを開きます。Digidata の各入出力チャンネルに Signal を作成します。

③Signal を作成する。Add ボタ①タブを選択する。

ンをクリックして Signal を新規に作成します。名前は任意で②チャンネルを選択する。

す。
Delete で削除、Rename で名前変更ができます。

④Scaling を設定する。

⑥必要であればソフトウェア
フィルタを設定します。
フィルタは RC１次です。

⑤Telegraph が設定してあると、自動的に Axopatch 200B の設定とGAIN、FREQUENCY、Cm が表示される。（この図では表示されていません。）

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

52
4.4.1. Lab Bench - Signal リスト
下表は作成する Signal のリストです。全部で７個の Signal を作成します。

Digitizer Channels
ANALOG IN 0

Signals
Im_scaled ( 0.001V/pA )Vm_scaled ( 0.001V/mV )ANALOG IN 1

10_Vm ( 0.01 V/mV )ANALOG IN 2

I_Output ( 0.001 V/pA )ANALOG IN 3

DNVO ( 1 V/V )ANALOG OUT 0

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

V_clamp ( 20 mV/V )I_clamp ( 2 nA/V )

必須
✔

✔

53
4.4.2. Lab Bench - Analog IN の設定
1.

Analog IN #0 (VC 用)VC モードで膜電流の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
②Analog IN #0 を選択する。

④0.001 V/pA に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③Im_scaled を作成する。

54
2.

Analog IN #0 (IC 用)

IC モードで膜電位の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
②Analog IN #0 を選択する。
③Vm_scaled を作成する。

④0.001 V/mV に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

55
3.

Analog IN #1

VC モードで膜電位の取り込みに使用する Signal を作成します。

①Input Signal を選択する。

②Analog IN #1 を選択する。

④0.01 V/mV に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③10_Vm を作成する。

56
4.

Analog IN #2

IC モードで膜電流の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
③I_Output を作成する。
②Analog IN #2 を選択する。

④0.001 V/pA に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

57
5.

Analog IN #3

VC-Patch モードでリセット信号を取り込むのに使用します。

①Input Signal を選択する。
③DNVO を作成する。

②Analog IN #3 を選択する。

④1 V/V に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

58
4.4.3. Lab Bench - Analog OUT の設定
1.

Analog OUT #0 (VC)

VC モードでコマンド電圧信号の出力に使用する Signal を作成します。

①Output Signal を選択する。
②Analog OUT #0 を選択する。
③V_clamp を作成する。

④20 mV/V に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

59
2.

Analog OUT #0 (IC)IC モードでコマンド電流信号の出力に使用する Signal を作成します。

①Output Signal を選択する。
②Analog OUT #0 を選択する。
③I_clamp を作成する。

④2 nA/V に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

60
4.5.

Membrane Test Setup の設定(Axopatch 200B)
Membrane Test とは記録を開始する前に、電極抵抗の測定や容量成分を補正するのに便利なオシロスコープ機能です。

①OUT 0 を選択する。
②Enable をチェックする。
③Im_scaled (pA)を選択する。
④V_clamp(mV)を選択する。
⑤OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

61
4.6.

Protocol の Channel 設定 (Axopatch200B)
Acquire/ New protocol を選択して、新規に Protocol を作成します。
下図のように９個のタブがあります。
アンプで個別に設定するのは、Inputs, Output タブにある Channel だけです。
LabBench で作成した Signal を Channel に設定します。
Inputs, Output 以外のタブについては、全機種で共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

62
4.6.1. VC モードの設定
3.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Im_scaled
Channel #1 → 10Vm

Channel #3 → DNVO
Channel #3 は 200B の Config
スイッチが Patch のときにのみ、オプションで使用します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

63
4.

Outputs タブ①Outputs タブを選択します。
②作成した Signal を設定します。
Channel #0 → V_clamp

VC モードなので単位が mV

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

64
5.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

65
4.6.2. IC モードの設定
1.

Inputs タブ

Acquire/ New protocol を選択して、新規に Protocol を作成します。

①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_scaled
Channel #2 → I_Output＊Channel 3 は使用しません。
オフにして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

66
2.

Outputs タブ①Outputs タブを選択します。
②作成した Signal を設定します。
Channel #0 → I_clamp

IC モードなので単位が nA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

67
3.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

68
4.7.

External Command の設定 (Axopatch200B)

External Command を有効にすると、Digidata からのコマンド信号が有効になります。
フロントパネルにある EXT. COMMAND スイッチを ON に設定します。
VC の場合は最大 200mV、IC の場合は最大 20nA まで印加することが可能です。
無効にする場合は OFF に設定します。
VC の場合、最大 1V まで使用することができます。
使用する場合は、下表のように接続を変更して、signal を追加します。また、Protocol の Output タブも変更する。
リアパネルにある EXT. COMMAND INPUT REAR SWITCH を ON に設定します。

Digitizer Channels
ANALOG OUT 0

Axopatch 200B の端子

Signals (VC)

Signals (IC)

EXT. COMMAND INPUT

V_clamp_x5

I_clamp

REAR SWITCH( 100 mV/V )( 2 nA/V )V_clamp ( 20 mV/V ) → V_clamp_x5 ( 100 mV/V )EXT. COMMAND INPUT FRONT SWITCH → EXT. COMMAND INPUT REAR SWITCH
これでセットアップは完了です。「12. Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

69
5.

Multiclamp 700B のセットアップ

5.1.

Mulriclamp700B の認識

1.

Multiclamp700B にヘッドステージを接続する。

2.

Multiclamp700B に電源ケーブルを接続する。

3.

Multiclamp700B と PC を USB1 ケーブルで接続する。（USB2 ケーブも使用可能）

4.

Multiclamp700B の電源を入れる。

5.

Multiclamp Commander 700B を起動する。

6. Select Device ツールボタン（）をクリックして Select Device ダイアログを開く。MultiClamp Hardware を選択して、Scan ボタンをクリックするとシリアル番号が表示されて認識する。
8.
9.

OK ボタンをクリックして終了する。
をクリックして Multiclamp 700B を初期化します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

70
5.2.

softpanel の認識

soft panel はつまみとスイッチで Multiclamp700B を操作するためのマニュアルコントローラです。（オプション品）

1. Select Device ツールボタン（）をクリックして Select Device ダイアログを開く。SoftPanel タブを選択して、Scan ボタンをクリックすると通信ポ

ート表示されて認識する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

71
5.3.

Multiclamp700B の接続

下表は Digidata との標準的な接続です。表に従って BNC ケーブルを接続して下さい。VC はボルテージクランプモード、IC はカレントクランプモードを意味します。
Digidata1550A / Digidata1550

Multiclamp700B の端子

用途

必須

ANALOG IN 0

PRIMARY OUTPUT 1

Channel 1 膜電流(VC) / 膜電位(IC)✔

ANALOG IN 1

SECONDARY OUTPUT 1

Channel 1 膜電位(VC) / 膜電流(IC)ANALOG OUT 0

COMMAND 1

Channel 1 コマンド 電位(VC) / 電流(IC)✔

ANALOG IN 2

PRIMARY OUTPUT 2Channel 2 膜電流(VC) / 膜電位(IC)✔

ANALOG IN 3

SECONDARY OUTPUT 2Channel 2 膜電位(VC) / 膜電流(IC)ANALOG OUT 0

COMMAND 2

Channel 2 コマンド 電位(VC) / 電流(IC)1440A /132xA の端子

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

✔

72
5.4.

Telegraphed Instruments の設定 (Multiclamp700B)

Telegraph とはアンプと pCLAMP の通信を行う機能です。pCLAMP はアンプから Gain やスケールなどの情報を受け取って、自動的に反映させます。
Configure/ Telegraph Instruments を選択して Telegraphed Instruments ダイアログを開きます。Digidata の各入力チャンネルに Telegraph の設定を行います。Telegraphed Instruments ダイアログの各項目を下表のように設定します。
1.

設定リスト

アンプ

Multiclamp 700B

Input Channel

Telegraph Instrument

Amplifier Output

Output Channel

ANALOG IN #0

Axon Multiclamp 700B

PRIMARY OUTPUT 1

ANALOG OUT #0

ANALOG IN #1

Axon Multiclamp 700B

SECONDARY OUTPUT 1

ANALOG OUT #0

ANALOG IN #2

Axon Multiclamp 700B

PRIMARY OUTPUT 2

ANALOG OUT #1

ANALOG IN #3

Axon Multiclamp 700B

SECONDARY OUTPUT 2

ANALOG OUT #1

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

73
2.

Analog IN #0 の設定①Digidata の入力チャンネルを選②アンプを選択する。

択します。

Axon Multiclamp 700B

Analog IN #0③Multiclamp 700B の出力端子を選択する。

④Multiclam700B のシリアル番号

Primary output 1

を選択する。選択できない場合はScan ボタンをクリックする。

⑤Multiclamp 700B のコマンド端子を選択する。
Analog OUT #0

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

74
3.

Analog IN #1 の設定①Digidata の入力チャンネルを選択します。

②アンプを選択する。
Axon Multiclamp 700B

Analog IN #1③Multiclamp 700B の出力端子を選択する。

④Multiclam700B のシリアル番号

Secondary output 1

を選択する。選択できない場合はScan ボタンをクリックする。

⑤Multiclamp 700B のコマンド端子を選択する。
Analog OUT #0

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

75
4.

Analog IN #2 の設定②アンプを選択する。
①Digidata の入力チャンネルを選

Axon Multiclamp 700B

択します。
Analog IN #2③Multiclamp 700B の出力端子を選択する。

④Multiclam700B のシリアル番号

Primary output 2を選択する。選択できない場合はScan ボタンをクリックする。

⑤Multiclamp 700B のコマンド端子を選択する。
Analog OUT #1

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

76
5.

Analog IN #3 の設定②アンプを選択する。
①Digidata の入力チャンネルを選

Axon Multiclamp 700B

択します。
Analog IN #3③Multiclamp 700B の出力端子を選択する。

④Multiclam700B のシリアル番号

Secondary output 2を選択する。選択できない場合はScan ボタンをクリックする。

⑤Multiclamp 700B のコマンド端子を選択する。
Analog OUT #1

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

77
5.5.

Lab Bench の設定 (Multiclamp 700B)
Configure/ Lab Bench を選択して Lab Bench ダイアログを開きます。Digidata の各入出力チャンネルに Signal を作成します。

①タブを選択する。

③Signal を作成する。Add ボタ
ンをクリックして Signal を新②チャンネルを選択する。

規に作成します。名前は任意です。

④Telegraph を設定したので、自動的に設定されグレー表示になる。

⑥必要であればソフトウェア
フィルタを設定します。
フィルタは RC１次です。

⑤自動的に Multiclamp 700B の設定とスケーリングが設定される。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

78
5.5.1. Lab Bench - Signal リスト
下表は作成する Signal のリストです。全部で１２個の Signal を作成します。

Digitizer Channels

Multiclamp700B

Signal (VC)

Signal (IC)必須

ANALOG IN 0

PRIMARY OUTPUT 1

Im_prime1

Vm_prime1✔

ANALOG IN 1

SECONDARY OUTPUT 1

Vm_sec1

Im_sec1

ANALOG OUT 0

COMMAND 1

Vclamp1

Iclamp1✔

ANALOG IN 2

PRIMARY OUTPUT 2

Im_prime2

Vm_prime2✔

ANALOG IN 3

SECONDARY OUTPUT 2

Vm_sec2

Im_sec2

ANALOG OUT 1

COMMAND 2

Vclamp2

Iclamp2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

✔

79
5.5.2. Lab Bench - Analog IN の設定
1.

Analog IN #0
Analog①
InputINSignals
#0 を選
択する。
タブを選択する。
Analog②
Analog
IN #0
INを選
#0
択する。
を選択する。
③ Im_prime1 を作成する。
④Vm_prime1 を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

80
2.

Analog IN #1
Analog①
InputINSignals
#0 を選
択する。
タブを選択する。
Analog②
Analog
IN #0
INを選
#1
択する。
を選択する。
③ Im_sec1 を 作成する。
④Vm_sec1 を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

81
3.

Analog IN #2

Analog①
InputINSignals
#0 を選
択する。
タブを選択する。
Analog②
Analog
IN #0
INを選
#2
択する。
を選択する。
③ Im_prime2 を作成する。
④Vm_prime2 を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

82
4.

Analog IN #3
Analog①
InputINSignals
#0 を選
択する。
タブを選択する。
Analog②
Analog
IN #0
INを選
#3
択する。
を選択する。
③ Im_sec2 を 作成する。
④Vm_sec2 を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

83
5.5.3. Lab Bench - Analog OUT の設定
1.

Analog OUT #0
Analog①
OutputIN #0
Signalsを選択する。
タブを選択する。
Analog IN #0②Analog
OUT
を選#0
択する。
を選択する。
③ Vclamp1 を 作成する。
④Iclamp1 を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

84
2.

Analog OUT #1
Analog①
OutputIN #0
Signalsを選択する。
タブを選択する。
Analog IN #0②Analog
OUT
を選#1
択する。
選択する。
③ Vclamp2 を 作成する。

④Iclamp2 を作成する。

⑤OK ボタンをクリックして終了します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

85
5.6.

Membrane Test Setup の設定 (Multiclamp700B)

Membrane Test とは記録を開始する前に、電極抵抗の測定や容量成分を補正するのに便利なオシロスコープ機能です。
5.6.1. Membrane Test Setup OUT 0 の設定①OUT 0 を選択する。
②チェックを入れる。
③単位が pA, nA など A のシグ
ナルを選択する。
④単位が mＶなど V のシグナルを選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

86
5.6.2. Membrane Test Setup OUT 1 の設定①OUT 1 を選択する。
②チェックを入れる。
③単位が pA, nA など A のシグ
ナルを選択する。
④単位が mＶなど V のシグナルを選択する。

⑤OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

87
5.7.

Protocol の Channel 設定 (Multiclamp700B)

Acquire / New protocol を選択して、新規に Protocol を作成します。下図のように９個のタブがありますが、アンプで個別に設定するのは、Inputs,
Output タブにある Channel だけです。Lab Bench で作成した Signal を Channel に設定します。Inputs, Output 以外のタブについては、共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

88
5.7.1. VC モードの設定
1.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Im_prime1
Channel #1 → Vm_sec1
Channel #2 → Im_prime2
Channel #3 → Vm_sec2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

89
2.

Outputs タブ①Outputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vclamp1
Channel #1 → Vclamp2

VC モードなので単位が mV

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

90
3.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

91
5.7.2. IC モードの設定
1.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_prime1
Channel #1 → Im_sec1
Channel #2 → Vm_prime2
Channel #3 → Im_sec2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

92
2.

Outputs タブ①作成した Signal を設定します。
Channel #0 → Iclamp1
Channel #1 → Iclamp2③ 必 要 に 応 じ て Analog OUT
Holding Levels を設定します。

④ 必 要 に 応 じ て Digital OUT
Holding Pattren を設定します。

⑤OK ボタンをクリックして終了します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

IC モードなので単位が nA93
3.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

94
5.8.

External Command の設定 (Multiclamp700B)

External Command を有効にすると、Digidata からの刺激信号が有効になります。
5.8.1. VC モードの設定
channel2 も同様に設定します。
Option ツールボタンをクリッ
クして、Option ダイアログを開きます。

External Command Sensitivity を 20mV/V に設定します。
最大 200mV までのコマンドを使用することができます。
１V まで使用したい場合は 100mV/V を選択します。
外部コマンドを使用しない場合は OFF に設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

95
5.8.2. IC モードの設定
channel2 も同様に設定します。
Feedback Resistor が 500MΩのとき、External Command
Sensitivity は 400pA/V か 2nA/V を選択することができます。
下図のように、Feedback Resister が変更されると、ExternalCommand Sensitivity も変更されます。

これでセットアップは完了です。「12. Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

96
6.

Axoclamp 900A のセットアップ

6.1.

Axoclamp 900A の認識

3.

Axoclamp 900A にヘッドステージを接続する。

4.

Axoclamp 900A に電源ケーブルを接続する。

5.

Axoclamp 900A と PC を USB2 ケーブルで接続する。（USB1 ケーブは使用不可）

6.

Axoclamp 900A の電源を入れる。

7.

Axoclamp Commander 900A を起動する。

8. Select Device ツールボタンをクリックして Select Device ダイアログを開く。AxoClamp Hardware を選択して、Scan ボタンをクリックするとシリア
ル番号が表示されて認識する。
10. OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

97
6.2.

Softpanel の認識
1. Select Device ツールボタンをクリックして Select Device ダイアログを開く。SoftPanel タブを選択して、Scan ボタンをクリックすると通信ポート
表示されて認識する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

98
6.3.

Axoclamp900A の接続
下の表は Digidata との標準的な接続です。表に従って BNC ケーブルで接続して下さい。
Digidata1550A / Digidata1550

Axoclamp 900A の端子

シグナル

ANALOG IN 0

Channel 1 SCALED OUTPUT

膜電位( dSEVC, TEVC, IC1, DCC )ANALOG IN 1

Channel 2 SCALED OUTPUTANALOG OUT 0

V-CLAMP COMMAND

コマンド 電位(dSEVC, TEVC )ANALOG OUT 1

Channel 1 I-CLAMP COMMAND

Channel 1 コマンド 電流( IC1 )ANALOG OUT 2

Channel 2 I-CLAMP COMMAND

Channel 2 コマンド 電流( IC2 )1440A /132xA の端子

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

膜電流( dSEVC, TEVC)
膜電位( IC2, HVIC)

99
6.4.

Telegraphed Instruments の設定 (Axoclamp900A)
Configure/ Telegraph Instruments を選択して Telegraphed Instruments ダイアログを開きます。Digidata の各入力チャンネルに Telegraph の設定を行います。Telegraphed Instruments ダイアログの各項目を下表のように設定して下さい。

アンプ

Input Channel

Telegraph Instrument

ANALOG IN #0

Axon Axoclamp 900A

ANALOG IN #1

Axon Axoclamp 900A

Axoclamp 900A

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Amplifier Output
Channel 1 SCALEDOUTPUT
Channel 2 SCALEDOUTPUT

Output Channel
ANALOG OUT #1
ANALOG OUT #2

100
1.

Analog IN #0 の設定①Digidata の入力チャンネルを選②アンプを選択する。

択する。

Axon Axonclamp 900A

Analog IN #0③Axoclamp 900A の出力端子を選択する。

④Axoclamp 900A のシリアル番号

Scaled output 1

を選択する。選択できない場合はScan ボタンをクリックする。

⑤Digidata の出力チャンネルを選択する。
Analog OUT #1

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

101
2.

Analog IN #1 の設定①Digidata の入力チャンネルを選択する。

②アンプを選択する。
Axon Axoclamp 900A

Analog IN #1③Axoclamp 900A の出力端子を選択する。

④Axoclamp 900A のシリアル番号

Scaled output 2を選択する。選択できない場合はScan ボタンをクリックする。

⑤Digidata のコマンド端子を選択する。
Analog OUT #2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

102
6.5.

Lab bench の設定 (Axoclamp 900A)

6.5.1. Lab Bench Signal リスト
下表は作成する Signal のリストです。全部で 9 個の Signal を作成します。
SignalSignalSignal(IC, DCC, HVIC)(dSEVC)(TEVC)

Channel 1 SCALED OUTPUT

Vm_1

Vm_dSEVC

Vm_1

ANALOG IN 1

Channel 2 SCALED OUTPUT

Vm_2

Im_dSEVC

Im_2

ANALOG OUT 0

V-CLAMP COMMAND－ANALOG OUT 1

Channel 1 I-CLAMP COMMAND

IC_Cmd1－－

ANALOG OUT 2

Channel 2 I-CLAMP COMMAND

IC_Cmd2－－

Digitizer Channels

Axoclamp 900A

ANALOG IN 0

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

VC_Cmd (20mV/V)

103
6.5.2. Lab Bench - Analog IN の設定
3.

Analog IN #0① Input Signals
タブを選択する。

② Analog IN #0
を選択する。

③Vm_1 を作成する。

④Vm_dSEVC を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

104
4.

Analog IN #1① Input Signals
タブを選択する。

② Analog IN #1#0を選択する。

③Vm_2 を作成する。

④Im_2 を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

105① Input Signals
タブを選択する。

② Analog IN #1
を選択する。

③ Im_dSEVC を作成する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

106
6.5.3. Lab bench - Analog OUT の設定
1.

Analog OUT #0① Output Signal を 選 択 す②Analog OUT #0 を選択する。

④20 mV/V に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③VC_Cmd を作成する。

107
2.

Analog OUT #1①Output Signal を選択す

②Analog OUT #1 を選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③IC_Cmd1 を作成する。

108
3.

Analog OUT #2①Output Signal を選択す

②Analog OUT #2 を選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③IC_Cmd2 を作成する。

109
6.6.

Protocol の Channel 設定 (Axoclamp900A)

Acquire/ New protocol を選択して、新規に Protocol を作成します。
下図のように９個のタブがあります。
アンプで個別に設定するのは、Inputs, Output タブにある Channel だけです。
Lab Bench で作成した Signal を Channel に設定します。
Inputs, Output 以外のタブについては、共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

110
6.6.1. TEVC モードの Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_1
Channel #1 → Im_2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

111
6.6.2. dSEVC モードの Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_dSEVC
Channel #1 → Im_dSEVC

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

112
6.6.3. TEVC, dSEVC モードの Outputs タブ①Outputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → V_Cmd③ 必 要 に 応 じ て Analog OUT

VC モードなのでHolding Levels を設定します。

単位が mV

④ 必 要 に 応 じ て Digital OUT
Holding Pattren を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

113
6.6.4. IC, DCC, HVIC モードの Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_1
Channel #1 → Vm_2

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

114
6.6.5. IC, DCC, HVI モードの Outputs タブ①作成した Signal を設定します。
Channel #0 → IC_Cmd1
Channel #1 → IC_Cmd2
IC モードなので③ 必 要 に 応 じ て Analog OUT
Holding Levels を設定します。

④ 必 要 に 応 じ て Digital OUT
Holding Pattren を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

単位が nA115
6.7.

External Command の設定 (Axoclamp900A)

External Command を有効にすると、Digidata からの刺激信号が有効になります。
6.7.1. TEVC, dSEVC モードの External Command①Option ツールボタンをクリックして、Option ダイアログを開きます。

②External Command Sensitivity を 20mV/V に設定します。
最大 200mV までのコマンドを使用することができます

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

116
6.7.2. IC, DCC, HVIC モードの External Command②External Command Sensitivity をチャンネルごとに設定することができます。スケーリングは使用するヘッドステージで異なります。
IC モードHS-9A x0.1

1 nA/Vmax ±10nA

HS-9A x1

10 nA/V

max ±100nA

HS-9A x10

100 nA/V

max ±1000nA

HS-9A x0.1

10 nA/V

max ±100nA

HS-9A x1

100 nA/V

max ±1000nA

HS-9A x101000 nA/V

max ±10000nA

HVIC モード

これでセットアップは完了です。「12. Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

117
7.

Axopatch 1D のセットアップ

7.1.

Axopatch 1D の設定

3.

Axopatch 1D にヘッドステージ、電源ケーブルを接続し、電源を入れる。

4.

Axopatch 1D を下表のように初期設定します。
設定項目

設定値

PIPETTE OFFSET

5.0

ZAP

0.5 msPIPETTE CAPACITANCE COMP.

最小（反時計回り）

SERIES RESISTANCE COMP. % PREDICTION

0 %, OFFSERIES RESISTANCE COMP. % CORRECTION

0 %, OFFSERIES RESISTANCE COMP. LAG

1 µs

WHOLE CELL CAP.

0 pF, OFFSERIES RESISTANCE

0 MΩ

HOLDING COMMAND

0 mV, x1, OFF

EXT COMMAND

OFF

METER

I

MODE

V-CLAMPCONFIG.

WHOLE CELL (β = 1)

OUTPUT GAIN:

α =1

LOWPASS BESSEL FILTER

5 kHz

LEAK SUBTRACTION∞ MΩ, OFF

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

118
7.2.

Axopatch 1D の接続

下表は Digidata との標準的な接続です。表に従って BNC ケーブルで接続して下さい。

Digidata の端子

Axopatch 1D の端子

ANALOG IN 0

SCALED OUTPUTANALOG IN 1

10 Vm

ANALOG IN 2

I

ANALOG OUT 0

EXT. COMMAND

TELEGRAPH INPUT 0

GAIN SETTING

GAIN 設定を取得

TELEGRAPH INPUT 1

FREQUENCY SETTING

Filter 設定を取得

ｓ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

用途
ボルテージクランプ時、膜電流の取り込み
カレントクランプ時、膜電位の取り込み

必須✔

ボルテージクランプ時、膜電位の取り込み
カレントクランプ時、膜電流の取り込み
ボルテージクランプ時、コマンド電位信号の出力
カレントクランプ時、コマンド電流信号の出力

✔
✔

119
7.3.

Telegraphed Instruments の設定 (Axopatch 1D)

Telegraph とはアンプと Clampex の通信を行う機能です。Clampex はアンプから Gain などの情報を受け取って、自動的に反映させます。
Configure/ Telegraph Instruments を選択して Telegraphed Instruments ダイアログを開きます。下図のように設定します。

①Digidata の入力チャンネルを②アンプを選択します。

選択します。

ヘッドステージによって設定がAnalog IN #0

異なります。
Axopatch-1 CV-4-1/100
Axopatch-1 CV-4-0.1/100③Digidata の Telegraph チャンネルを選択します。
Gain → Telegraph Inputs 0Frequency→ Telegraph Inputs 1

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

120
7.4.

Lab bench の設定 (Axopatch 1D)

7.4.1. Lab Bench - Signal リスト
下表は作成する Signal のリストです。全部で 6 個の Signal を作成します。Configure/ Lab Bench を選択して Lab Bench ダイアログを開きます。
Digidata の各入出力チャンネルに Signal を作成します。

Digitizer Channels

Axopatch 200B の端子

ANALOG IN 0

SCALED OUTPUTANALOG IN 1

10Vm

ANALOG IN 2

I

ANALOG OUT 0

EXT. COMMAND

Signals (VC)

Signals (IC)Im_scaled (*1)

Vm_scaled (*1)

( 0.001V/pA )( 0.001xβ V/mV )10_Vm( 0.01 V/mV )－－I_Output( 0.001 V/pA )V_clamp (*2)

I_clamp (*2)

( 20 mV/V )( 200÷β pA/V )(*1) ANALOG IN 0 は２つの Signal を作成する必要があります。Axopatch 1D の SCALED OUTPUT 端子は、VC モードの時は膜電流、IC モードの時は膜電位を出力するからです。
(*2) ANALOG OUT0 は２つの Signal を作成する必要があります。EXT. COMMAND 端子は、VC モードの時はコマンド電位、IC モードの時はコマン
ド電流のシグナルを入力する必要があるからです。
βはヘッドステージゲインです。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

121
7.4.2. Lab Bench - Analog IN の設定
1.

Analog IN #0 (VC 用)VC モードで膜電流の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
②Analog IN #0 を選択する。

④0.001 V/pA に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③Im_scaled を作成する。

122
2.

Analog IN #0 (IC 用)

IC モードで膜電位の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
②Analog IN #0 を選択する。
③Vm_scaled を作成する。

④0.001 V/mV に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

123
3.

Analog IN #1

VC モードで膜電位の取り込みに使用する Signal を作成します。

①Input Signal を選択する。

②Analog IN #1 を選択する。

④0.01 V/mV に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③10_Vm を作成する。

124
4.

Analog IN #2

IC モードで膜電流の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
③I_Output を作成する。
②Analog IN #2 を選択する。

④0.001xβ V/pA に設定する。
β=0.1 のとき 0.0001 V/pA
β=1 のとき 0.001 V/pA
β=100 のとき 0.1V/pA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

125
7.4.3. Lab Bench - Analog OUT の設定
4.

Analog OUT #0 (VC)

VC モードでコマンド電圧信号の出力に使用する Signal を作成します。

①Output Signal を選択する。
②Analog OUT #0 を選択する。
③V_clamp を作成する。

④20 mV/V に設定する。
（ 1D の EXT. COMMANDSENSITIVITY を 1mV/V に設定する場合は、1mV/V に設定する。
）

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

126
5.

Analog OUT #0 (IC)IC モードでコマンド電流信号の出力に使用する Signal を作成します。

①Output Signal を選択する。
②Analog OUT #0 を選択する。
③I_clamp を作成する。

④200÷βpA/V に設定する。
β=0.1 のとき 2n A/V
β=1 のとき 200 pA/V
β=100 のとき 2 pA/V
（ 1D の EXT. COMMANDSENSITIVITY を 1mV/V に設定する場合は、10÷βpA/Vに設定する。
）
β=0.1 のとき 100p A/V
β=1 のとき 10pA/V
β=100 のとき 0.1 pA/V

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

127
7.5.

Membrane Test Setup の設定(Axopatch 1D)
Membrane Test とは記録を開始する前に、電極抵抗の測定や容量成分を補正するのに便利なオシロスコープ機能です。

①OUT 0 を選択する。
②Enable をチェックする。
③Im_scaled (pA)を選択する。
④V_clamp(mV)を選択する。
⑤OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

128
7.6.

Protocol の Channel 設定 (Axopatch 1D)
Acquire/ New protocol を選択して、新規に Protocol を作成します。
下図のように９個のタブがあります。
アンプで個別に設定するのは、Inputs, Output タブにある Channel だけです。
LabBench で作成した Signal を Channel に設定します。
Inputs, Output 以外のタブについては、全機種で共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

129
7.6.1. VC モードの設定
6.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Im_scaled
Channel #1 → 10Vm

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

130
7.

Outputs タブ①Outputs タブを選択します。
②作成した Signal を設定します。
Channel #0 → V_clamp

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

131
8.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

132
7.6.2. IC モードの設定
9.

Inputs タブ

Acquire/ New protocol を選択して、新規に Protocol を作成します。

①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_scaled
Channel #2 → I_Output

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

133
10. Outputs タブ①Outputs タブを選択します。
②作成した Signal を設定します。
Channel #0 → I_clamp

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

134
11. Protocol の保存
Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

135
7.7.

External Command の設定 (Axopatch 1D)

External Command を有効にすると、Digidata からのコマンド信号が有効になります。Axopatch-1D の EXT. COMMAN SENSITIVITY スイッチを
20mV/V に設定します。無効にする場合は OFF に設定します。
微小電圧を使用する場合は、Axopatch-1D EXT. COMMAN SENSITIVITY スイッチ 1mV/V に設定します。また、LabBench で Signal のスケーリング
を 1mV/V に変更する必要もあります。

20 -> 1

これでセットアップは完了です。「Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

136
8.

Axoclamp 2B のセットアップ

8.1.

Axoclamp 2B の設定

Axoclamp 2B にヘッドステージ、電源ケーブルを接続し、電源を入れる。Axoclamp 2B を下表のように初期設定します。
設定項目

設定値

Rate

最小

STEP COMMAND

0OFF/EXT./CONTスイッチ

OFF

DESTINATIONスイッチ

任意

Anti –Alias

最小

Gain

最小

Phase Lag

0.01

Phase Multiplier

OFF

Holding Position

5

Capacitance Neutralization

最小BRIDGE0

INPUT OFFSET

5OUTPUT OFFSET

5

DC CURRENT COMMANDOFF

OUTPUT BANDWIDTH30kHz

I DISPLAYIm

H1

ME1のヘッドステージゲインHに合わせる

H2

ME2のヘッドステージゲインHに合わせる

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

137
8.2.

Axoclamp 2B の接続
下の表は Digidata との標準的な接続です。表に従って BNC ケーブルで接続して下さい。

Digidata の端子

Axoclamp 2B の端子

ANALOG IN 0

10Vm

ME1 膜電位の取り込み

BRIDGE, DCC, SEVCANALOG IN 1Im

ME1 膜電流の取り込み

BRIDGE, DCC, SEVCANALOG IN 2

V2

ME2 膜電位の取り込み

BRIDGEANALOG IN 3

I2

ME2 膜電流の取り込み

BRIDGE, TEVCANALOG IN 4

0.1 x I2

ME2 膜電流の取り込み（TEVC 大電流用）

TEVC

ANALOG OUT 0

EXT. VC COMMAND

ボルテージクランプのコマンド電位信号を出力

SEVC, TEVCANALOG OUT 1

EXT. ME1 COMMAND

カレントクランプの ME1 コマンド電流の出力

BRIDGE, DCC

ANALOG OUT 2

EXT. ME2 COMMAND

カレントクランプの ME2 コマンド電流の出力

BRIDGE, DCC

用途

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

使用するモード

138
8.3.

Lab bench の設定 (Axoclamp 2B)

8.3.1. Lab Bench Signal リスト
下表は作成する Signal のリストで、LabBench で作成します。使用するモードによって作成する Signal が異なります。H はヘッドステージゲインです。
次のページより実際に Lab Bench で作成していきます。

Digitizer Channels

SignalScalling

使用するモード

ANALOG IN 0

Vm_1

0.01 V/mV

BRIDGE ME1, DCC, SEVCANALOG IN 1

Im_1

(10÷H) mV/nA

BRIDGE ME1, DCC, SEVCANALOG IN 2

Vm_2

0.001 V/mV

BRIDGE ME2ANALOG IN 3

Im_2

(10÷H) mV/nA

BRIDGE ME2, TEVCANALOG IN 4

Im_2_TEVC(1÷H) mV/nA

TEVC

ANALOG OUT 0

Cmd_VC

20 mV/V

SEVC, TEVCANALOG OUT 1

Cmd_ME1(10 x H) nA/V

BRIDGE ME1, DCC

ANALOG OUT 2

Cmd_ME2(10 x H) nA/V

BRIDGE ME2, DCC

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

139
8.3.2. Lab Bench - Analog IN の設定
Lab Bench Signal リストのシグナルを下図のように作成します。Add をクリックして新しいシグナルを作成します。

H が 0.1 のとき (10÷H) mV/nA -> 100 mV/nA -> 0.1V/nA
H が 1 のとき (10÷H) mV/nA -> 10 mV/nA -> 0.01V/nA
H が 10 のとき (10÷H) mV/nA -> 1 mV/nA -> 0.001V/nA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

140

H が 0.1 のとき (10÷H) mV/nA -> 100 mV/nA -> 0.1V/nA
H が 1 のとき (10÷H) mV/nA -> 10 mV/nA -> 0.01V/nA
H が 10 のとき (10÷H) mV/nA -> 1 mV/nA -> 0.001V/nA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

141

日本語 OS はμが文字化けします

H が 0.1 のとき (1÷H) mV/nA -> 10 mV/nA -> 0.01V/nA
H が 1 のとき (1÷H) mV/nA -> 1 mV/nA -> 0.001V/nA
H が 10 のとき (1÷H) mV/nA -> 0.1 mV/nA -> 0.0001V/nA -> 0.1V/uA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

142
8.3.3. Lab bench - Analog OUT の設定
Lab Bench Signal リストのシグナルを下図のように作成します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

143

H が 0.1 のとき (10 x H) nA/V -> 1 nA/V

H が 0.1 のとき (10 x H) nA/V -> 1 nA/V

H が 1 のとき (10 x H) nA/V -> 10 nA/V

H が 1 のとき (10 x H) nA/V -> 10 nA/V

H が 10 のとき (10 x H) nA/V -> 100 nA/V

H が 10 のとき (10 x H) nA/V -> 100 nA/V

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

144
8.4.

Protocol の Channel 設定 (Axoclamp 2B)

Acquire/ New protocol を選択して、新規に Protocol を作成します。
下図のように９個のタブがあります。
アンプで個別に設定するのは、Inputs, Output タブにある Channel だけです。
Lab Bench で作成した Signal を Channel に設定します。
Inputs, Output 以外のタブについては、共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

145
8.4.1. TEVC モードの Channel 設定

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

146
8.4.1. SEVC モードの Channel 設定

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

147
8.4.2. BRIDGE, DCC モードの Channel 設定

DCC モードは ME1 のみ使用可能

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

148
8.5.

External Command の設定 (Axoclamp2B)

External Command は常に有効です。
これでセットアップは完了です。「12. Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

149
9.

OC-725C のセットアップ

9.1.

OC-725C の接続
下の表は Digidata との標準的な接続です。表に従って BNC ケーブルで接続して下さい。
Digidata1550A / Digidata1550
1440A /132xA の端子
ANALOG IN 0
ANALOG IN 1
ANALOG OUT 0
TELEGRAPH INPUT 3

OC-725C の端子
I MONITOR(BATH ELECTRODES section)
Vm×10(VOLTAGE ELECTRODES section)

COMMAND IN ÷10(COMMANDS section)

GAIN TELEGRAPH OUTPUT(on rear panel)

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

シグナル

膜電流
膜電位
コマンド電位GAIN の同期

150
9.2.

Telegraphed Instruments の設定 (OC-725C)

Telegraph とはアンプと Clampex の通信を行う機能です。Clampex はアンプから Gain などの情報を受け取って、自動的に反映させます。
Configure/ Telegraph Instruments を選択して Telegraphed Instruments ダイアログを開きます。下図のように設定します。

①Digidata の入力チャンネルを②アンプを選択します。

選択します。

Warner OC-725C

Analog IN #0③Digidata の Telegraph チャンネルを選択します。
Gain → Telegraph Inputs 3

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

151
9.3.

Lab bench の設定 (OC-725C)
Configure/ Lab Bench を選択して Lab Bench ダイアログを開きます。Digidata の各入出力チャンネルに Signal を作成します。

9.3.1. Lab Bench Signal リスト
下表は作成する Signal のリストです。全部で 3 個の Signal を作成します。

Digitizer Channels

OC-725C

Signals

I MONITOR

I_oc725c(BATH ELECTRODES section)( 1V/μA )Vm×10

V_oc725c(VOLTAGE ELECTRODES section)( 0.01 V/mV )COMMAND IN ÷10

VC_oc725c(COMMANDS section)

100mV/V

ANALOG IN 0
ANALOG IN 1
ANALOG OUT 0

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

152
9.3.2. Lab Bench - Analog IN の設定
12. Analog IN #0
膜電流の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
②Analog IN #0 を選択する。

④1V/μA に設定する。
日本語 OS は「μ」が「ｵ」に文字化けするので注意してください。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③I_oc725c を作成する。

153
13. Analog IN #1
膜電位の取り込みに使用する Signal を作成します。

①Input Signal を選択する。

②Analog IN #1 を選択する。

④0.01 V/mV に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③V_oc725c を作成する。

154
9.3.3. Lab Bench - Analog OUT の設定
刺激電圧の出力に使用する Signal を作成します。

①Output Signal を選択する。
②Analog OUT #0 を選択する。

④ 100mV/V に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③VC_oc725c を作成する。

155
9.4.

Membrane Test Setup の設定(OC-725C)
Membrane Test とは記録を開始する前に、電極抵抗の測定や容量成分を補正するのに便利なオシロスコープ機能です。

①OUT 0 を選択する。
②Enable をチェックする。
③I_oc725c(μA)を選択する。
④VC_oc725c(mV)を選択する。
⑤OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

156
9.5.

Protocol の Channel 設定 (OC-725C)

Acquire/ New protocol を選択して、新規に Protocol を作成します。
下図のように９個のタブがあります。
アンプで個別に設定するのは、Inputs, Output タブにある Channel だけです。
LabBench で作成した Signal を Channel に設定します。
Inputs, Output 以外のタブについては、共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

157
1.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → I_oc725c
Channel #1 → V_oc725c

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

158
2.

Outputs タブ①Outputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → VC_oc725c③ 必 要 に 応 じ て Analog OUT

VC モードなのでHolding Levels を設定します。

単位が mV

④ 必 要 に 応 じ て Digital OUT
Holding Pattren を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

159
3.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。
これでセットアップは完了です。「12. Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

160
10. MEZ-7200 & CEZ-1200 のセットアップ
10.1. MEZ-7200 & CEZ-1200 の接続
下の表は Digidata との標準的な接続です。表に従って BNC ケーブルで接続して下さい。

CEZ-1200 の端子

MEZ-7200 の端子

シグナル

Vm

A – B NON FILTER

膜電位

CEZ-1200 の端子

シグナル

ANALOG IN 0

Im

膜電流（ボルテージクランプ用）

ANALOG OUT 0

PULSE Ⅰ

コマンド電位（ボルテージクランプ用）

MEZ-7200 の端子

シグナル

ANALOG IN 1

A–B

膜電位（電極抵抗測定用・ボルテージクランプ用）

ANALOG IN 2

CURRENT

膜電流（電極抵抗測定用）

ANALOG OUT 1

STIMULUS INPUT

コマンド電流（電極抵抗測定用）

Digidata1550A / Digidata1550
1440A /132xA の端子

Digidata1550A / Digidata1550
1440A /132xA の端子

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

161
10.2. Lab bench の設定 (MEZ-7200 & CEZ-1200)
Configure/ Lab Bench を選択して Lab Bench ダイアログを開きます。Digidata の各入出力チャンネルに Signal を作成します。
10.2.1. Lab Bench Signal リスト
下表は作成する Signal のリストです。全部で 5 個の Signal を作成します。

Digitizer Channels

CEZ-1200

Signals
Im_ CEZ
10 -4 のとき ( 0.01 V/μA )ANALOG IN 0

Im

10 -5 のとき ( 0.1 V/μA )10 -6 のとき ( 0.001 V/nA )10 -7 のとき ( 0.01 V/nA )10 -8 のとき ( 0.1 V/nA )VC_ CEZANALOG OUT 0

PULSE Ⅰ

x1 のとき ( 1 V/V )x1/10 のとき ( 100 mV/V )x1/100 のとき ( 10 mV/V )Im のスケーリングは SENSITIVITY(A/V) つまみの設定で異なります。
PULSE Ⅰのスケーリングは PULSE INPUT つまみの設定で異なります。PULSE INPUT つまみは最大コマンド電圧で決定します。x1 は最大 10V、
x1/10 は最大１V、x1/100 は最大 100mV です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

162

Digitizer Channels

MEZ-7200ANALOG IN 1

A–B

SignalsVm_ MEZ( 1 V/V )Im_ MEZANALOG IN 2

CURRENT

JZ-101J のとき ( 0.01 V/nA )JZ-102J のとき ( 0.1 V/nA )IC_ MEZANALOG OUT 1

STIMULUS

JZ-101J のとき( 10 nA/V )JZ-102J のとき ( 1 nA/V )STIMULUS のスケーリングは STIMULATION ダイアルの設定によって異なります。上記は１回転の場合のスケールです。回転数を掛け算したスケーリングになります。JZ-101J のとき 10 回転の場合は 100 nA/V、0.1 回転の場合は 1 nA/V になります。
STIMULATION ダイアルはどのように決めればよいか？例えば、電極抵抗を測定する場合は電流の大きさで決めます。MEZ-7200 のマニュアルを見ると、電圧測定範囲は 2V(2000mV)です。測定できる電極抵抗の範囲は電流の大きさできまります。
2000mV / 1nA= 2000MΩ （JZ-102J で STIMULATION を 0.1 回転に設定）
2000mV / 10nA= 200MΩ （JZ-101J で STIMULATION を 0.1 回転に設定、JZ-102J で STIMULATION 1 回転に設定）
2000mV / 100nA = 20 MΩ （JZ-101J で STIMULATION を 1 回転に設定、JZ-102J で STIMULATION 10 回転に設定）
2000mV / 1000nA = 2MΩ （JZ-101J で STIMULATION を 10 回転に設定）

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

163
10.2.2. Lab Bench - Analog IN の設定
1.

Analog IN #0

膜電流の取り込みに使用する Signal を作成します。

①Input Signal を選択する。
③Im_CEZ を作成する。
②Analog IN #0 を選択する。

④SENSITIVITY(A/V) つまみに合わせてスケーリングを設定する。
10 -4 のとき 0.01 V/μA
10 -5 のとき 0.1 V/μA
10 -6 のとき 0.001 V/nA
10 -7 のとき 0.01 V/nA
10 -8 のとき 0.1 V/nA
日本語 OS は「μ」が「ｵ」に文字化けするので注意してください。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

164
10.2.3. Analog IN #1
膜電位の取り込みに使用する Signal を作成します。

①Input Signal を選択する。

②Analog IN #1 を選択する。

④1 V/V に設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③Vm_ MEZ を作成する。

165
10.2.4. Analog IN #2
膜電流（電極抵抗測定用）の取り込みに使用する Signal を作成します。

①Input Signal を選択する。

②Analog IN #2 を選択する。

④プローブに合わせてスケーリングを設定する。
JZ-101J のとき 0.01 V/nA
JZ-101J のとき 0.1 V/nA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

③Im_ MEZ を作成する。

166
10.2.5. Lab Bench - Analog OUT の設定
1.

Analog OUT #0

コマンド電圧の出力に使用する Signal を作成します。

①Output Signal を選択する。
③VC_CEZ を作成する。
②Analog OUT #0 を選択する。

④ PULSE INPUT つまみに合わせてスケーリングを設定する。
x1 のとき 1 V/V
x1/10 のとき 100 mV/V
x1/100 のとき 10 mV/V

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

167
10.2.6. Analog OUT #1
コマンド電流（電極抵抗測定用）の出力に使用する Signal を作成します。

①Output Signal を選択する。
③IC_MEZ を作成する。
②Analog OUT #1 を選択する。

④プローブと STIMULATION つ
まみの設定に合わせてスケーリングを設定します。
STIMULATION つまみが 1 回転
JZ-101J のとき

10 nA/V

JZ-102J のとき 1 nA/V
STIMULATION つまみが 10 回転
JZ-101J のとき

100 nA/V

JZ-102J のとき 10 nA/V

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

168
10.3. Membrane Test Setup の設定(MEZ-7200 & CEZ-1200)
Membrane Test とは記録を開始する前に、電極抵抗の測定や容量成分を補正するのに便利なオシロスコープ機能です。（動作未確認です。使用できないかもしれません）①OUT 0 を選択する。
②Enable をチェックする。
③Im_CEZ を選択する。
④VC_CEZ を選択する。
⑤OK ボタンをクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

169
10.4. Protocol の Channel 設定 (MEZ-7200 & CEZ-1200)
Acquire/ New protocol を選択して、新規に Protocol を作成します。
下図のように９個のタブがあります。
アンプで個別に設定するのは、Inputs, Output タブにある Channel だけです。
LabBench で作成した Signal を Channel に設定します。
Inputs, Output 以外のタブについては、共通項目なので別章で解説します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

170
10.4.1. ボルテージクランプの設定
1.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Im_CEZ
Channel #1 → Vm_MEZ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

171
2.

Outputs タブ①Outputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → VC_CEZ③ 必 要 に 応 じ て Analog OUT
Holding Levels を設定します。

④ 必 要 に 応 じ て Digital OUT
Holding Pattren を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

VC モードなので
単位が V

172
3.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

173
10.4.2. カレントクランプの設定（電極抵抗測定用）
1.

Mode/Rate タブ

MEZ-7200 を使用して、電極抵抗を測定するプロト
コルを作成します。プロトコルの詳細については「Protocol の基本設定」の章を参考にして下さい。
①Mode を Episodic stimulation に設定します。

②Sweep duration を 0.05s に設定します。

③サンプリング周波数を 10000Hz(10kHz)に設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

174
2.

Inputs タブ①Inputs タブを選択します。

②作成した Signal を設定します。
Channel #0 → Vm_MEZ
Channel #1 → Im_MEZ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

175
3.

Outputs タブ①Outputs タブを選択します。

②作成した Signal を設定します。
Channel #1 → IC_MEZ

IC モードなので単位が nA

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

176
4.

Math タブMath は入力波形の演算を行う機能です。入力した電圧と電流から抵抗値を計算します。

①Math タブを選択します。

②シグナルを設定します。

③ゲインとオフセット設定します。

④演算を設定します。

⑤演算式が表示されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

177
5.

Waveform タブの設定

コマンド電流を作成します。

②図のように設定する。

③コマンド波形を作成する。

①Channel1 を選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

178
下図のようなコマンド電流を作成しました。

10nA
20ms

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

179
記録すると、下図のように電圧、電流に加えて、演算された抵抗値の波形も表示されます。

抵抗値の波形

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

180
6.

Protocol の保存

Acquire / Save Protocol As を選択します。Save Protocol As ウィンドウが表示されるので、ファイル名を入力して保存ボタンをクリックします。

①ファイル名を入力する。
②ファイルの種類を選択する。デフォルトで選択されている「Pro」が protocol の拡張子です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

181
10.5. External Command の設定 (MEZ-7200 & CEZ-1200)
CEZ-1200 は MODE スイッチを CLAMP に設定すると、Digidata からのコマンド信号が有効になります。
ボルテージクランプするとき、CEZ-1200 のフロントパネルにある MODE スイッチを CLAMP に設定して下さい。
MEZ-7200 は、Digidata からのコマンドが常に有効になっています。
電極測定のとき、MEZ-7200 は特に設定する必要はありません。
しかし、CEZ-1200 のフロントパネルにある MODE スイッチを OFF に設定して下さい。

これでセットアップは完了です。「12. Protocol の基本設定」の章でプロトコルの設定を行って下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

182
11. Protocol の設定（基本編）
11.1. Protocol の作成
Acquire/ New protocol を選択して、新規に Protocol を作成します。すでに作成した Protocol を再編集する場合は、Open Protocol で既存の Protocol
を開き、Acquire/ Edit Protocol で再編集します。
その他コマンド
Save Protocol：再編集した Protocol を上書き保存する。
Save Protocol As：新規に作成した Protocol や再編集した Protocol を、別名で新規に保存する。
ツールバーにショートカットがあります。

Open

Save

ProtocolProtocolClampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Save
ProtocolAs

Edit
Protocol183
11.2. Mode/Rate タブの設定
11.2.1. Gap-free モード
Gap-free モードは長時間の取り込みができます。また、刺激波形が使用できないなど、機能に制限があります。

①Mode/rate タブを選択する。
②Gap-free を選択します。

③Duration を選択して測定時間を設定します。
Use available disk space を選択した場合「ｵ」は「μ」が文字化けしています。
100μs

は、ハードディスクがいっぱいになるま
で取り込みを行います。停止したい場合
は STOP コマンドを使用します。

④Sampling rate per signal にサンプリン
グ周波数を設定します。

データ容量が表示さ

サンプリング周波数とは１秒間に取り込れます。

むポイント数です。
大きいほど速い反応を測定することができ、データ容量は大きくなります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

184

刺激波形を設定する Waveform タブは無効になり、刺激波形とデジタル制御が使用できません。

Holding Level は設定可能です。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

185

オプションの刺激波形を設定する
Stimulus タブは無効になります。

統計機能を設定する Statistics タブは無効になります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

186
11.2.2. Episodic stimulation モード
Episodic stimulation モードはステップパルスなどの刺激波形を加えることができます。

①Mode/rate タブを選択する。

②Episodic stimulation を選択する。
③Trial Hierarchy を設定する。
Trial delay：測定開始までの待ち時間
Run/trial：繰り返し数
Sweep/Run：Sweep の数
Sweep Duration：Sweep の時間(次項の図をご参考下さい)④Fast Rate にサンプリング周波数を設定する。

⑤Start-to-Start Interval の Sweep を設定します。次の Sweep までの時間です。
Minimum に設定すると、最小時間が設定されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

データ容量が表示されます。

187
Trial, Run, Sweep の関係

スタート

測定開始

Trial delay

Sweep

Sweep2

Sweep3durationStart-to-Start
Interval Sweep
Run2

Run1
Start-to-Start Interval Run

Trial

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

188
最大ポイント数
1Sweep 当たりで記録できるサンプル数には限界があり、最大 1032258 まで記録することができます。また、サンプル数は Sweep
duration と Fast Rate で決まります。（サンプル数＝Sweep duration×Fast Rate）

しかし、実際に刺激波形が設定できるのは Epoch と呼ばれる領域で、最大 1000000 ポイントです。これは Sweep の前後に First holding、Last Holding と呼ばれる時間が存在するからです。この時間は避けることができません。
First

Last

Holding

Holding

Epoch
Sweep Duration
刺激波形が設定できるのはこの領域 Epochs

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

測定開始

189
11.3. Input タブの設定
入力チャンネルを設定するタブで、使用するチャンネルをチェックし、Lab Bench で作成したシグナルを選択します。下図は Axopatch200B VC モードの場合の例です。詳細は各機器の「Protocol の Channel 設定」の章をご参照下さい。

①使用するチャンネルをチェックします。

②Lab Bench で作成したシグナルを選択します。Channel #3 DNVO は Axopatch 200B がPatch モードの時に、オプションで使用します。通常は必要ありません。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

190
11.4. Output タブの設定
出力チャンネルを設定するタブで、Lab Bench で作成したシグナルを選択します。また、アナログ出力の Holding Ｌｅｖｅｌとデジタル出力の Holding
Pattern を設定します。下図は Axopatch200B VC モードの場合の例です。詳細は各機器の「Protocol の Channel 設定」の章をご参照下さい。

①アナログ出力の Signal を選択します。

②アナログ出力の Holding Level を設定します。

③デジタル出力の Holding Pattern を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

191
11.5. Trigger タブの設定
トリガーを設定するタブで、測定モードによって設定できる項目が異なります。通常はデフォルト設定のままにします。
11.5.1. Trigger タブの設定(Gapfree モード)

①Acquire > Record もしくは （）で記録を開始した後、実際に記録を開始するタイミングを設定します。通常は Immediate に設定します。
即座に記録開始するモードです。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

192
11.5.2. Trigger タブの設定(Episodic Stimulation モード)

①Acquire > Record もしくは （）で記録を開始した後、実際に記録を開始するタイミングを設定します。通常は Immediate に設定します。
即座に記録開始するモードです。

②記録を開始するトリガー信号を設定します。通常は Internal Timer に設定します。Digidata の内部トリガーに同期して記録を開始するモードです。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

193
11.6. Waveform タブの設定
Waveform タブは刺激波形を設定するタブです。アナログ波形は Epoch を使用する方法と、ファイルを読み込む方法があります。デジタル信号の設
定は Epoch を使用する方法のみです。ここでは例として Epoch の設定を行います。

②Analog Waveform をチェックします。
③波形の作成方法は Epochs を選択します。
④Intersweep holding level は Useholding を選択します。
⑤Epoch に刺激波形を設定します。

⑥Update Preview をクリックす
ると、刺激波形を確認できます。

⑦他のチャンネルを設定する。
⑧OK をクリックして終了する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

194
Epoch の設定
例として簡単なステップパルスを作成します。
EpochA を設定します。

EpochB を設定します。

波形タイプを選択します。

1Sweep 目（Sweep0）のレベルを設定します。
1Sweep 目（Sweep0）の時間を設定します。

2Sweep 目以降のレベル増減を設定します。
2Sweep 目以降の時間増減を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

195
Epoch のレポート

Sweep 数が表示される。

選択している Epoch のステータスが表示されます。
Final Level は最後の Sweep のレベルを表示します。
この例では-100mV から始まり、最後の 10Sweep 目は80mV になること表示しています。
First duration は最初の Sweep の時間を表示します。
こ の 例 で は 、 Delta Duration が ゼ ロ な の で Final
Duration はレポートされません。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Epoch の使用率がレポート
されます。

196
Waveform Preview の表示

Epoch B の Final Level
Delta Level

Epoch A の First Level
Epoch B の First Level
Sweep
EpochＡ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

EpochB

197
11.7. Protocol の保存
Acquire/ Save Protocol As を選択して、Protocol を新規に保存します。再編集した Protocol を上書き保存する場合は、Save Protocol を選択します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

198
12. 細胞へアプローチする(Membrane Test)
細胞へアプローチするとき、Membrane Test を使用すると便利です。Membrane Test は電極抵抗の測定や容量成分を補正するのに便利なオシロ
スコープ機能です。ここでは、例としてホールセル実験の流れをみていきます。

Membrane Test を開始する

BATH
PATCH

BATH 溶液に電極をつける

ギガシ－ル状態にする
ベースラインを調整する

CELL
ホールセル状態にする

電極容量を補正する
電極容量を補正する

膜容量を補正するシリーズ抵抗を補正する

Membrane Test を停止する

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

199
12.1. Membrane Test Setup の設定
Configure/ Membrane Test Setup を選択して、Membrane Test Setup ダイアログを開きます。下図は Axopatch200B VC モードの場合の例です。
アンプによって設定がことなりますので、各アンプの設定の章をご確認下さい。設定が完了したら、Tools/ Membrane Test…を選択して、Membrane
Test を起動します。

①Digidata の出力チャンネルを選択する。
②Enable をチェックする。
③入力チャンネルの signal を選択する。
＊signal はアンプによって異なります!
④出力チャンネルの signal を選択する。
＊signal はアンプによって異なります!
⑤膜容量の計算範囲を設定する。
次項の図をご参考下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

200
膜容量の計算範囲設定した範囲で膜容量を計算する。
ここでは、80% ~ 20%の範囲で膜容量を計算している。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

201
12.2. BATH 状態
電極を BATH 溶液に入れた状態でパルス電圧を印加します。今回は Patch-1U モデルセルの BATH をヘッドステージに接続します。BATH には電極抵抗を模擬した 10MΩの直列抵抗が入っています。

#0 を選択します。

電極容量によってトランジェントが発生する。

BATH を選択します。
Rt：電極抵抗
今回はモデルセルの 10MΩが表示される。

パルスの振幅、周波数、Holding を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

202
アンプの Pipette Offset でベースラインを調整し、Pipette Capacitance Compensation で電極容量によるトランジェントを補正します。

Pipette Capacitance Compensation によってトランジェントが削除された。

モニターする Analog OUT を選択します。
Digidata1550：#0 - 7
Digidata1440A：#0 - 3
Digidata1322A：#0 - 1
Hold：保持電流（平均電流）( 振幅 － ベース ) ÷ 2
Averaging をチェックすれば、平均波形で計算することができます。平均する回数を設定します。
Tracking をチェックすれば、自動でオフセットを補正すことができます。速度を設定します。

Pipette Offset でベースラインを調整する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

203
12.3. PATCH 状態
細胞にアプローチして、ギガシール状態にします。今回は Patch-1U モデルセルの PATCH をヘッドステージに接続します。PATCH にはギガシールを模擬した 10GΩの抵抗が入っています。

BATH 状態では見えなかったトランジ
ェントが発生する。
PATCH を選択します。
ギガシール状態を表示する。

BATH と個別に設定しておくことができます。好みによりますが、PATCHのときに holding 電圧を設定する人もいます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

204
再度、アンプの Pipette Capacitance Compensation を使用して電極容量によるトランジェントを補正します。

Pipette Capacitance Compensation によってトランジェントが削除された。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

205
12.4. CELL 状態
細胞膜を破り、ホールセル状態にします。今回は Patch-1U モデルセルの CELL をヘッドステージに接続します。CELL には電極抵抗を模擬した 10MΩと膜抵抗を模擬した 500MΩの直列抵抗が入っています。さらに、膜容量を模擬した 33pF のコンデンサが 500MΩと並列にはいっています。

膜容量によって遅いトランジェン
CELL を選択します。
様々なパラメータを表示します。
Cm：膜容量Rm：膜抵抗
Ra：直列抵抗（電極抵抗など）
Tau：時定数
Hold：保持電流

BATH、PATCH と個別に設定しておくことができます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

トが発生する。

206
アンプの Whole-Cell Capacitance Compensation を使用して膜容量によるトランジェントを補正します。必要であればシリーズ抵抗も補正して下さい。これで、Membrane Test による前準備は終了です。Membrane Test を終了して記録を開始します。

Membrane Test で測定した Cm と Ra を参考にして、Whole-Cell

CapacitanceCompensation を行います。

Whole-Cell Capacitance Compensation
によってトランジェントが削除された。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

207
13. 記録する
13.1. 保存先の設定
File/ Set Data File Names を選択して、保存先とファイル名を設定します。

Date prefix を無効にすると、任意のファイル名を設定できる。

Data prefix を有効にすると、日付ベースのファイル名を設定できる。
Allow long file names を無効にすると短いファイル名を設定できます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

208
13.2. 記録の開始①Acquire/ Open Protocol を選択して、Protocol を開きます。

②Acquire/ Recode を選択して、記録を開始します。もしくは Acquisition ツールバーの Record をクリックします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

209③その他の Acquire/ xxxxx コマンド
Re-Recode：停止するまで（stop を選択するまで）記録を繰り返す。
View Only：表示して、記録はしない。
Pause View：記録を中断して、表示を行います。再開する場合は再度 Pause View を選択します。Episodic モードでは使用できません。
Pause：記録を中断します。再開する場合は再度 Pause を選択します。Episodic モードでは使用できません。
Stop：記録、表示を停止する。
Repeat：有効にすると、記録や表示を繰り返す。
White Last：最後に表示したデータを保存する。
④Acquisition ツールバー

RepeatRe-Recode

RecodeClampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Pause
View

View

Stop

210
14. Protocol の設定（応用編）
14.1. ハムノイズを除去する（Digidata1550A, Digidata1550B のみ）
Digidata1550A と Digidata1550B はアナログ入力にハムノイズ除去機能が搭載されています。Edit Protocol > Input タブの Hum Silencer をチェックすると有効になります。また、リアルタイムコントロールバーには設定項目が表示されます。Adapt をチェックするとノイズパターンを読み込み、Subtractをチェックすると、ノイズをキャンセルし、Clear をクリックすると、ノイズパターンをリセットします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

211【注意】Adapt を有効にして、HumSilencer ノイズキャンセルでレコーディングを始めるとき、HumSilencer システムが初期設定する間、21ms レコーディ
ングの開始遅延があるかもしれない。
【注意】シグナルに大きなステップやスパイクがある場合は、Adapt を選択してはいけません。because fast changes will be visible as an excursion
scaled to -1/50 of the fast signal amplitude in the output、また、ノイズレプリカの繰り返し平均を継続することによって、電源周波数を得ているからです。Episodic stimulation モードのとき、Adapt は自動的にチェックがはずされます。

測定データには HumSilencer のコメントタグが挿入されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

212
Gap-free, Fixed-length events, High-speed oscilloscope, Variable-length events モードのとき
1. Edit Protocol > Inputs タブで Analog IN Channel #0 の HumSilencer をチェックする。
2. 選択されていなければ、リアルタイムコントロールパネルの Subtract を選択して、ノイズキャンセルを有効にします。
Adapt は自動的にチェックされます。
3. Record か View Only をクリックして測定を開始され、測定中に HumSilencer の設定は変更可能です。
Episodic stimulation モードのとき
1. Edit Protocol > Inputs タブで Analog IN Channel #0 の HumSilencer をチェックする。
2. 選択されていなければ、リアルタイムコントロールパネルの Subtract を選択して、ノイズキャンセルを有効にします。
Subtract を選択しないと、ノイズキャンセルは無効になります。
3. Record か View Only をクリックして測定を開始すると、Adapt チェックボックスは自動的にはずされ、Subtract チェックボックスは手順２のままです。測定中、HumSilencer コントロールパネルは変更できません。
【注意】HumSilencer は Pre-sweep Train, P/N Leak Subtraction, Membrane Test Between Sweeps が機能している時間もノイズをキャンセル
します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

213
すべての測定モードにおいて、HumSilencer で修正されたデータは通常のシグナルと異なる色で表示することができます。Scope ウィンドウの View
> Window Properties > Signals から色を設定することができます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

214
14.2. Epoch ごとにサンプリングレートを変更する（Gear Shift）
Mode/Rate タブの Slow rate をチェックすると、Waveform タブの Epoch ごとに Fast rate と Slow rate のどちらかを選択することができます。例えば、重要な Epoch のみ高速レート、その他は低速レートで記録することにより、ファイルサイズを小さくすることができます。

①Slow rate をチェックする。

②Epoch の Sample rateに Fast と Slow のどち
らかを選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

215
14.3. 平均波形を記録する

Run/trial を２以上にすると、Start-to-Start Interval の Run が有効になります。Run の開始から次の Run の開始までの時間を設定します。

Run/trial を２以上にすると、Averaging が有効になり、平均波形を記録します。
注意）Run を２以上にすると、常に平均波形を記録するので、同じ Protocol を２回以上繰り返す場合は、シーケンス機能を使用します。
使用方法は「Sequencing Key」の章で説明します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

データ容量が表示されます。

216
平均値の計算方法を選択する。
Option ボタンをクリックすると、Averaging Options ウィンドウが表示されます。平均値の計算方法と Undo File を設定します。Undo Fileについては後述します。

累積の平均値を計算します。ドリフトが少ないときに選択して下さい。
例）
Run/trial = 3 のとき
Avg1=Run1
Avg2=(Run1+Run2) / 2
Avg3=(Run1+Run2+Run3) / 3

重みを設定した平均値を計算します。ドリフトするときに選択して下さい。Weighting に重みを設定します。
例）
Run/trial = 3 のとき
Avg1=Run1
Avg2=(Avg1 × Weighting ) + Run2 × (100 - Weighting)
Avg3=(Avg2 × Weighting ) + Run3 × (100 - Weighting)

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

217
Undo File を設定する。

Always prompt to allow undo は記録終了後、Save Average
ウィンドウが表示され、何番目の平均波形を保存するか選択します。Update every には保存する頻度を設定します。

Prompt on premature termination only は記録を途中で停止した場合のみ、Save Average ウィンドウが表示され、何番目の平均波形を保存するか選択します。

表示している平均波形の番号を表示します。
Undo File とはUndo File は Run ごとに作成される平均波形を保存する機能です。最終的にデータとして保存する平均波形を Save Average ウィンドウで選択することができます。
よって、最適な平均波形を選択することができます。

何番目の平均波形を保存するか選択します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

218
14.4. さまざまな測定モード
14.4.1. Fixed-length events モード
Fixed-length events モードは波形と threshold レベルの交点をトリガーとして、トリガー前後のデータを記録するモードです。必要な部分のみ記録することができるので、ファイルサイズを小さくできます。また、トリガー前後の記録時間を前と後を別に設定できします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

219
データ取得モードを Fixed-length events モードに設定しますと、下図のような設定画面となります。
Trial Length：データ測定時間を設定します。
Use available disk space：ディスク容量の許す限り測定
Duration(hh:mm:ss) データを測定する時間
Sampling：サンプリングを設定します。
Sampling Interval per Signal(Hz)：サンプリングレートを設定
Sweep length per event per signal(s)： データ記録時間

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

220
Trigger タブを選択すると、下図の画面が表示されます。Threshold レベルと測定時間を Trigger Setting に設定します。
Pre-trigger length (ms) ：トリガー前の記録時間を設定
Trigger threshold ：threshold レベルを設定
Polarity：トリガーの動作する方向を設定
Rising 立ち上がり
Falling 立ち下がり

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

221
14.4.2. Variable-length events モード
Variable-length events は波形と threshold レベルの交点をトリガーとして、トリガー前後のデータを記録するモードです。必要な部分のみ記録することができるので、ファイルサイズを小さくできます。また、トリガー前後の記録時間は前と後で同じです。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

222
データ取得モードを Variable-length events モードに設定しますと、下図のような設定画面となります。
Trial Length：データ測定時間を設定します。
Use available disk space：ディスク容量の許す限り測定
Duration(hh:mm:ss) データを測定する時間
Sampling：サンプリングを設定します。
Sampling Interval per Signal(Hz)：サンプリングレートを設定

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

223
Trigger タブを選択すると、下図の画面が表示されます。Threshold レベルと測定時間を Trigger Setting に設定します。
Pre-trigger length (ms) ：トリガー前の記録時間を設定
Trigger threshold ：threshold レベルを設定
Polarity：トリガーの動作する方向を設定
Rising 立ち上がり
Falling 立ち下がり

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

224
14.4.3. High-speed oscilloscope モード
High-speed oscilloscope モードはオシロスコープにように動作するモードです。

Trial Length：データ測定時間の設定
Use available disk space：ディスク容量の許す限り測定
Duration(sweep)：何画面分データを取り込むか設定
Sampling
Sampling Interval per Signal(Hz)：サンプリングレートを設定
Sweep length per event signal(s)：一画面で表示するサンプ
ル数を設定

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

225
14.5. トリガー信号の応用
14.5.1. Scope Trigger を使用する
Digidata の同期信号の端子が測定中に high になります。端子名は Digidata によってことなります。また、Episodic モードのみは Sweep duration（First Holding から End Holding の間）で high なります。

①Scope trigger をチェックする。

端子名
ADC CLOCK OUTPUT (Digidata 1320/1)
TRIGGER OUTPUT BNC (Digidata 1322)
SCOPE BNC (Digidata 1440A, 1550).

Episodic モードの場合
刺激信号
Seep
duration
同期信号

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

226
14.5.2. 外部トリガーに同期してコメントを挿入する
コメントの詳細については「測定中にコメントを挿入する」の章をご参照下さい。

タグの種類
Time：コメントなしのタグ
Comment：コメント入力できるタグ
Voice：音声を録音できるタグ① Digidata の Tag 端子に外部の同期信号を入力する②External tags をチェックする。
③タグの種類を選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

227
14.5.3. 外部トリガーに同期して統計を計算する① Digidata の START 端子に同期信号を入力する。

③ Threshold-based statistics source を
Digitizer START Input に設定する。

② Threshold-based statistics をチェ ックする。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

228
14.5.4. 外部トリガーに同期して記録を開始する①Digidata の START 端子に外部の同期信号を入力する。
②Start trial width を Digitizer START Inputに設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

229
14.5.5. 外部トリガーに同期して、Sweep, Run を開始する①Digidata の START 端子に外部の同期信号を入力する。

②Trigger source を Digitizer START Inputに設定する。
③Action に同期する対象を選択する。

Mode/Rate タブの Start-to-Start
Interval は無効になる。

External tags は使用できなくなる。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

230
14.6. 統計機能を使用する
14.6.1. Gapfree モード
Gapfree モードは Trigger タブで統計機能を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

231
14.6.2.

Stimulus, High-speed oscilloscope モード

Statistics タブに統計機能を設定します。結果は Online Statistics ウィンドウにレポートされます。

⑨計算項目を選択します。
①入力信号を選択する。
②極性を設定する。

③ベースラインの範囲を設定する。
④計算の番号を選択する。最大８箇所まで設定可
能です。
⑤計算範囲を設定する。
⑥フィルタを設定する。1-21 まで設定でき、数値が大きいほどノイズを除去します。通常は１に設定します。
⑦記録終了後、保存する場合はチェックします。
⑧保存後、Online Statistics ウィンドウをクリアする
場合はチェックします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

232
Online Statistics ウィンドウ

波形が表示される。

数値が表示される。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

233
14.7. コメント機能を使用する
14.7.1. ファイルにコメントを付ける①White each trial を選択する。

②温度、圧力、濃度などの数値を入力する。
③コメントを入力する。
④アンプの測定モードを選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

234

Prompt each trial を選択すると、記録終了後に Comments ウィンドウが表示されます。記録の度にコメントを修正できます。

記録終了後に自動的に表示される。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

235
14.7.2. 測定中にコメントを挿入する
測定中にコメントを挿入する機能で、メニューの Acquire から選択できます。Time Tag は単純なマーキング、Comment Tag はコメント付
マーキング、Voice Tag は音声録音付マーキングを挿入する機能です。
ショートカットキーを使うと便利です。

コメントを入力します。タグを入れた時点で記録されます。あわててコメントを入力する必
要はありません。

録音停止
タグ無効
録音延長
100%になると自動的に終了します。延長したい場合は Extendをクリックして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

236
メニューの Acquire > Voice Tags から Voice tag の設定をすることができます。

①ノイズを除去する機能で、設定したスレッシュホール
ドより小さい音声は除去されます。
②録音した音声を圧縮形式で保存します。ファイルサイズは小さくなりますが、記録の最後に遅れが発生します。
スレッシュホールドが表示されます。

③自動的に終了するまでの時間を設定します。
④外部トリガーによる Voice Tag を設定します。上は外部トリガーが Low から High への変化を検出して、VoiceTag Duration で設定した時間だけ録音するモードです。
下は外部トリガーが High になっている限り録音を続けるモードです。
外部トリガーによる Voice Tag は Protocol >Trigger タブ
で有効にできます。
「外部トリガーに同期してコメントを挿入する」の章をご参照下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

237

Voice Tag

右クリック、Play で再生できます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Comment Tag

Tag

238
14.8. 入力信号を演算する
Math Signal は２つの入力信号を演算する機能です。

①入力信号を選択します。
②General purpose を選択します。
③演算を選択します。

④ゲイン、オフセットの係数を設定します。

演算式がレポートされます。

⑤演算結果の表示範囲と単位を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

239

演算波形が追加されます。
設定した単位が表示されます。
設定した範囲が表示されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

240
Ratio dyes は比率の演算を行うモードで、蛍光分析測定のキャリブレーションなどに使用します。

①Ratio dyes を選択します。

②Ratio にする前の信号のオフセットを設定します。
Ratio の演算式がレポートされます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

241
14.9. デジタル信号の設定（Digital Output）
Digital Output 端子からトリガー信号を出力して、電磁弁やスティミュレータなど周辺機器を同期することができます。
14.9.1. 基本設定
② Digital Outputs を チ ェ ッ ク し ま す 。
Intersweep bit pattern には sweep 間のパ
タ ー ン を 設 定 し ま す 。 Use holding は
Output タブの Digital OUT Holding Patternを使用し、Use last epoch は最後の Epochのパターンを使用します。
③デジタル信号を２進数で設定します。
1010 → High, Low, High, Low
#3

#2

#1 #0

0000 → Low, Low, Low, Low
#7

#6 #5

#4

①デジタル信号を同期するアナログ出
力チャンネルを選択します。同期できるチャンネルは１つだけです。
④Update Preview をクリックすると、デジタル信号を確認できます。
Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

242⑤ デジ タル信号の 初期値を 設 定 する。チェックすると high になります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

243
左から順に

左から順に

#3 #2 #1 #0#3 #2 #1 #0#7 #6 #5 #4#7 #6 #5 #4

左から順に

左から順に

Intersweep
Use holding

Sweep１

Sweep2Start-to-Start Interval Sweep

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Use last epoch

244

1010

0000

0

1

1

0

0

1

1

0

0

1

0

1

0

1

0

1

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

0101

1111

245
14.9.2. 交互にデジタル信号を出力する
sweep0 は設定値、sweep1 は全て Low, sweep2 は設定値、sweep3 は全て Low, …のように、「設定値」と「全て Low」を交互に出力することができます。

Alternate Digital Outputs をチェックする。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

246
14.9.3. 連続したトレインを出力する①使用するチャンネルを「＊」に設定します。
②Train rate に周波数、Pulse width にパル
ス幅を設定します。

③周波数を設定すると、パルス数と 10%のパルス幅がレポートされます。パルス数とパルス幅を設定する参考にして下さい。

④Active high logic for digital trains のチェックをはずすと、出力は反転します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

247

Active high logic for digital trains を OFF にした場合

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

248
14.10.

いろいろな波形を作成

14.10.1. 複数のステップパルスを作成する

Epoch を追加する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

249
14.10.2. 時間にデルタを設定する

Delta duration を設定する。

Delta duration

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

250
14.10.3. ランプ波形を作成する

Ramp に設定する。

EpochB First level

EpochB

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

251

Ramp を連続する。

EpochB First level

EpochC First level

EpochB

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

EpochC

252
14.10.4. トレイン波形を作成する

Pulse train を選択する。

周波数のパルス幅を設定する。

パルスの設定がレポートされる。

Train Rate

Pulse width

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

253
トレイン波形の種類

Biphasic train

Triangle train

Cosine train

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

254
14.10.5. 記録データなどを刺激波形に使用する（Stimulus File）
Waveform タブの Analog Waveform で Stimulus file を選択すると、記録データなどを刺激波形に使用することができます。

①Stimulus file を選択する。

②Stimulus File にファイルを選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

255③ファイルを設定します。フォーマットは abf,
atf, dat の３つです。
④チャンネル、Sweep、ゲイン、オフセットを設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

256

アナログ信号の設定は無効になります。

⑤デジタル信号は Epoch で設定できます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

257
14.10.6. ATF 形式で Stimulus File を作成する
ATF( Axon Text File )形式で刺激波形を作成し、Stimulus File として設定することが可能です。ATF 形式はタブ区切りのテキストファイルなので、テキストエディタやエクセルで編集できます。
１行目にファイル形式とバージョンを指定する。

２行目にオプションヘッダー行数を指定する。オプションヘッダー情報は不要なので「0」とします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

258
２行目に使用する列数（Sweep 数 + 1）を指定する。「+1」は時間データ分です。

Sweep 数は Mode/Rate タブの
Sweep/run で決まります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

259
３行目に列の名前を指定します。１列目は時間、２列目以降は刺激の信号です。信号名は任意で、単位は不要です。

４行目以下に時間と刺激値を指定します。時間の単位は「s」です。刺激値の単位は Analog Output の Signal で決まります。

刺激値の単位は Analog Outputの Signal で決まります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

260
時間をプロトコルに合わせる必要があります。時間間隔は Fast rate、時間範囲は Sweep Duration で決まります。

0.1995 - 0.1994 = 0.0001s = 0.1ms = 100μs

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

261
タブ区切りの TXT 形式で保存し、ファイル名の変更で拡張子を ATF 形式に変更する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

262
Stimulus File をプロトコルにセットします。

①Stimulus file を選択する。

②Stimulus File にファイルを選択する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

263③ファイルの種類を ATF に設定して、ファイルを選択する。
④Sweep、ゲイン、オフセットを設定します。通常は図のようにデフォルト設定を使用します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

264
Update preview で期待通りの波形が作成されているか確認しておきましょう。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

265
14.11.

オプション波形14.11.1. 刺激波形の前にトレイン波形を追加する（Pre-Sweep Train）
Stimulus タブの Pre-Sweep Train はメイン Sweep の前に連続した矩形波を印加する機能です。矩形波を設定する Train 領域、Holding Level を設定
する Post-train 領域に分かれています。メイン Sweep の First Holding Level は Post-train になります。

スタート

Step duration

Baseline duration
Train
Holding

Post-train duration

Pre-Sweep Train

また、Real Time Control パネルには Sweep の数とともに、Pre-Sweep が表示されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

メイン Sweep

266
パルス周波数と時間がレポートされる。
パルス数
Baseline の時間
Baseline の振幅
Step の時間

Post-train の時間

Step の振幅

Post-train の振幅

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

267
14.11.2. リーク電流を除去する（P/N Leak Subtraction）
Stimulus タブの P/N Leak Subtraction はサブ Sweep を印加することによって細胞のリーク電流を測定し、メイン Sweep による反応からリーク電流を減算する機能です。Waveform タブの Analog waveform が有効になっている場合に使用できます。サブ Sweep の時間はメイン Sweep と同じで、振幅は縮小されます。縮小される度合いはサブ Sweep の波形数によって決まります。（サブ Sweep の振幅＝ メイン Sweep の振幅 / サブ Sweep 数）

時間は同じ

・・・・・・・････････････・・・・・・・・・・・・・・・

振幅は縮小

サブ Sweep

メイン Sweep1

サブ Sweep

メイン Sweep2

サブ Sweep の反応は clampex で見ることができません。確認したい場合はオシロスコープや他の Digitizer（Mini-Digi など）を使用する必要があります。また、Real Time Control パネルには Sweep の数とともに、サブ Sweep の数も表示されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

268
サブ Sweep の挿入箇所をメ
イン Sweep の前後に設定する。複数の Analog OUT を使
P/N Leak Subtraction を適用する入力チ用している場合は、全てのチ

ャネルのシグナルを設定します。

ャンネルにおいて同じ設定とサブ Sweep の極性なります。

を設定する。

サブ Sweep の数を設定します。「サブSweep の振幅＝ メイン Sweep の振幅 /サブ Sweep 数」となります。また、サブ
Sweep の数をメイン Sweep 数ごとに設定する場合は User List を使用します。詳細は User List を参照して下さい。
サブ Sweep とサブ Sweep の間隔を設定する。
サブ Sweep の前後待ち時間を設定する。
記録開始からサブ Sweep 開始までの時
間と、サブ Sweep 終了からメイン Sweep
開始までの時間（Pre-Sweep を設定している場合は Pre-Sweep 開始までの時間）

チェックするとサブ Sweep の反応を
減 算 さ れ た デ ー タ を 表 示 し 、 RealTime Control パネルには Corrected と表示される。チェックをはずすと減算されていない生データが表示され、Real Time Control パネルには Not
Corrected と表示される。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

記録開始からサブ
Sweep 開始までの時間の Holding Level 設
定する。

269

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

270
14.11.3. Sweep ごとに時間やレベルを設定する（User List）
Stimulus タブの User List はメイン Sweep の Start-to-Start と Epoch、Pre-Sweep の全設定項目、P/N Leak Subtraction のサブ Sweep 数をカス
タマイズする機能です。

カ ス タ マ イズ す る
項目を選択する。
設定値を入力する。

例１）Pre-Sweep Train のパルス数を Sweep ごとに設定する。

Number of pulse intrain はグレー表示になります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

271例２）Epoch A の Level を Sweep ごとに設定する。

Epoch は List と表示されます。

例３）Epoch A のデジタル信号を Sweep ごとに設定する。

Epoch は List と表示されます。

注意）List of parameter value に設定した数が Sweep 数とより少ない場合は、最後の設定値を使用します。設定した数が Sweep 数より多い場合は、無視しますClampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

272例４）２連続パルスの間隔を Sweep ごとに設定する。
例として、Epoch B の duration を Sweep ごとに設定します。

設定項目を設定する
設定値を設定する

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

273

User List の設定値で決まります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

274
Waveform Preview には Sweep1 しか表示されませんが、設定した波形が出力されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

275
出力波形を記録した結果は下図のようになります。設定した波形が反映されているのがわかります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

276例５）マルチパルス

Digital bit pattern を使用するのでEpisodic Stimulation を選択します。
刺激セットの周期を設定します。Sweep duration を長くする必要があります。

刺激セットの数を設定します。1,2,4,8,16 回刺激の場合は５セットなので ” 5 “に設定します。
Sweep の時間は（刺激周期）×（最大刺激数）より長くする必要があります。
パルス周期が 100Hz(10ms)のとき、
160ms より大きくする必要があります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

277

Digital outputs を有効にします。

Type を Pulse に設定します。
使用するチャンネルを”＊”に設定します。

パルス周期を設定します。100Hz = 10ms

パルス幅を設定します。パルス周期より短くする必要があります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

278

User List で設定する項目を選択します。今回は Epoch A
duration を選択します。

Epoch A duration の値を設定します。今回は Sweep 数が 5なので、Sweep1, Sweep2, …Sweep 5 のようにコンマで区切って設定します。
パルス周期を 100Hz = 10ms の場合、Epoch A duration が 10ms のとき、1 回刺激
Epoch A duration が 20ms のとき、2 回刺激・・
Epoch A duration が 160ms のとき、16 回刺激となります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

279
14.11.4. Sweep 間に Membrane Test を挿入する（Membrane Test Between Sweeps）
Stimulus タブの Membrane Test Between Sweeps は Sweep と Sweep の間に Membrane Test を行う機能です。

Sweep の終了と Membrane Test

Membrane Test 終 了 か ら 次 の開始までの時間

Sweep 開始までの時間

また、Mode/rate タブの Start-to Start Interval を設定する必要があります。Minimum では使用できません。

・・・・・・・････

Sweep1

Settling time

Membrane

Settling time

Before

Test

After

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Sweep

280
Membrane Test Between Sweeps のサンプル数を選択することができます。Configure / Membrane Test Setup の Membrane Test Between
Sweeps セクションで選択します。

Sweep ごとに 1 サンプルする。

Membrane Test のサンプリング設定でサンプルする。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

281

Single measurement per sweep

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Use Membrane Test setting

282
15. エクスポート機能
15.1.1. clampex9 のプロトコルをエクスポートする

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

283
15.1.2. データを pCLAMP9（ABF1.8）にエクスポートする

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

284
15.1.3. 複数のデータを pCLAMP9（ABF1.8）にエクスポートする
Data File Index を作成して、abf ファイルをグループ化し、まとめてエクスポートします。File > New File Data Index を選択して、Add Existing Files
ダイアログを開きます。Browse をクリックしてグループ化するデータフォルダを選択します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

285
Data File Index が作成されますので、エクスポートしたいファイルを選択します。右クリックして Export Files を選択します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

286
File Export ダイアログが開きます。出力フォルダ、File Type を選択して、OK をクリックします。

出力先を設定Data Format を設定

ファイルヘッダー名を追加できます

Data File Index にData Format が 記載されています。同
じ Data Format を選択して下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

287
必要であれば、File > Save Data File Index As で Data File Index ファイルを保存します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

288
16. シーケンス機（Sequencing Keys）
Configure/ Sequencing Keys を選択して下さい。
16.1. 記録を繰り返す①Add ボタンを押して新規に Sequencing Key を作成します。

②使用するショーカットキーを選択します。
OK をクリックします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

289③Protocol ボタンをチェックします。
④「Action」に動作を設定します。
「Record」を選択します。
⑤「protocol file」に使用する Protocol を設定します。
Browse ボタンを押して選択します。
⑥「Repetition count」に繰り返し数を設定します。
⑦「Repetition start-to-start」に繰り返し時間を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

290⑧「Next key」は「None」に設定します。
⑨Ok をクリックして終了します。
「Ctrl」キー「1」キーを同時に押してみて下さい。
「Repetition count」に設定した回数だ
け繰り返して記録を行うはずです

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

291
16.2. 異なるプロトコルを連結して記録する

例として、Ctrl+2 と Ctrl+3 を連動するシーケンスを作成します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

292

Ctrl+3 を選択する。

記録するプロトコルを設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

293

None を選択する。

記録するプロトコルを設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

294

Ctrl+2 と Ctrl+3 が連動したシーケンスが作成できました。
「Ctrl」キー「2」キーを同時に押してみて下さい。Ctrl+2 と Ctrl+3 が連続して記録されるはずです。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

295
17. ツール機能
17.1. Comment to Lab Book
Tool > Comment to Lab Book は System Lab Book ウィンドウにコメントを挿入する機能です。System Lab Book ウィンドウは操作や発生したイ
ベントを記録するテキストエディタです。System Lab Book ウィンドウは Comment to Lab Book を使用せずに直接入力することもできます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

296
Configure > Lab Book Option は Lab Book のオプションを設定します。

記録するイベントを設定します。
Never log any events
イベントは記録しません。
Log vital events
主要イベントのみ記録します。
Log all events

イベントはテキストファイルで保存されています。ファイルを設定できます。

全てのイベントを記録します。
bak 拡張子のバックアップを作成する
日時を記録するオプションです。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

オプションです。

297
17.2. Zero Stopwatch
Tools > Zero Stopwatch はストップウォッチの時間をリセットする機能です。ストップウォッチはテータスバーに表示されています。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

298
17.3. Membrane Test を中断してトレイン波形を印加する

パルスを設定する。
Number of pulse in train
パルス数
Pre-train duration
パルス前の待ち時間
Baseline duration
ベースラインの時間
Baseline level
ベースラインのレベル
Step duration
パルスの時間Step level
パルスのレベル
Post-train duration
パルス後の待ち時間

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

パルスの周波数とトレインの時間がレポートされる。

299

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

300

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

301
17.4. Junction Potential
Junction Potential は細胞内外の液間電位を計算するツールです。計算した液間電位を clampex またはアンプに設定して、細胞内外の液間電位を補正した実験が行えます。
17.4.1. Junction Potential の使い方
例として、以下の条件で設定を行います。


実験：ホールセルパッチクランプ



温度：20℃

リファレンス電極：Standard salt solution electrode（アガーブリッジなど）



バス溶液





140 mM NaCl

2.8 mM KCl



2 mM MgCl2



1 mM CaCl2

0 mM HEPES



PH7.4 調整のために NaOH を追加（5 mM HEPES-, 5 mM Na+）電極溶液
145 mM K-gluconate

8 mM NaCL



1 mM MgCl2



10 mM HEPES



PH7.4 調整のために NaOH を追加（5 mM HEPES-, 5 mM Na+）

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

302
Tools > Junction Potential を選択して、Calculate Junction potential を起動する。New Experiment を選択して、New Experiment ダイアログを表示する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

303

実験のタイプを選択して、Next をクリックする。

温度を設定して、Ｎｅｘｔをクリックする。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

パッチクランプのタイプを選択して、Next をクリックする。

リファレンス電極のタイプを選択して、Next をクリックする。

304

実験手順が表示される。

実験手順が表示される。

電極を BATH 溶液に入れたら Pipette Offset でゼロに調整ししばらくすると、細胞内液と電極溶液が完全に混ざって液間て下さい。

電位 VLi はゼロになる。Offset は VL’のみになります。

Next をクリックする。

Finish をクリックする。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

305
実験の設定が完了したので、次にイオンの設定を行います。Add をクリックして Add Ion Concentration ダイアログを開きます。Na イオンのイオン濃度を設定して、Ok をクリックします。

分子を選択する。

電極溶液の分子濃度を設定する。
バス溶液の分子濃 度を設定する。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

306
Na イオンの設定が表示されます。同様に、K, Cl, Gluc, Mg, Ca, HEPES のイオン濃度も設定します。

液間電位が計算される。

任意のイオンを追加できます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

307
Save をクリックして設定を保存できます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

308
Load をクリックして設定を開くことができます。その際、Query ダイアログが表示されるので、イオンリストと実験タイプなど全ての設定を開く場合はYes、イオンリストのみを開く場合は No を選択して下さい。

実験や温度は異なっても、イオンリストが同じ場合に使用します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

309
17.4.2. 液間電位を設定する。
ボルテージクランプの場合 Clampex に設定する場合は、Acquire > Edit Protocol の Outputs タブに設定します。 または、clampex の右側にあるReal Time Controls Panel に設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

310
Multiclamp 700B Commander に設定する場合は、V-Clamp タブの Holding に設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

311
カレントクランプの場合
Clampex に設定する場合は、Acquire > Edit Protocol の Inputs タブに設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

312
Multiclamp 700B Commander に設定する場合は、I-Clamp タブの Pipette Offset に設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

313
17.5. LTP Assistant
LTP Assistant は LTP プロトコルを簡単に作成することができるツールです。メニュー > Tools > LTP Assistant を選択して下さい。
【注意】
LTP Assistant にはバグが発生する場合があります、それを回避するために設定が複雑になる場合があります。LTP Assistant を使用せずに、個別にプロトコル作成して、Sequencing Key でプロトコルを連結することをおすすめ致します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

314
17.5.1. LTP プロトコルの作成
New をクリックすると、New Experiment ダイアログが開きます。名前とフォルダを設定して、OK をクリックします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

315
17.5.2. Sequencing の設定
各キーの設定をします。まずは Ctrl+1 を選択して、Properties ボタンをクリックします。

Next Key は本来 Ctrl+2 に設定すべきですが、バグ回避のために None に設定します。
Action は View か Record を選択します。
Duration はバグ回避のために（Protocol time には間違った数値が設定されています）Sequencing time を選択して、時間を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

316
Ctrl+2, Ctrl+3 についても同様に Next key, Action, Duration を設定します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

317
17.5.3. Input/Output チャンネルの設定
チャンネルの設定を行ないます。

刺激波形のチャンネルを設定します。
Analog OUT #1 と Digital OUT#1 を使用する場合は、設定しAnalog OUT #0 の signal と Holdingを設定します。

Analog IN の Signal を設定します。
Analog OUT と Digital OUT の合計で2 チャンネル以上使用する場合は、
Analog IN チャンネルも２チャンネル
以上有効にして下さい。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

ます。

318
17.5.4. Baseline プロトコルの設定

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

319

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

320

警告が表示される
Waveform Stimulation はPresynaptic Stimulation より長い時間に設定する必要があります。短いと右上に警告が表示されます。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

321
17.5.5. Conditionig プロトコルの設定

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

322

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

323
Close をクリックすると、変更を保存するか聞かれるので、Yes をクリックします。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

324
17.5.6. Sequencing key の設定
次に、LTP Assistant での設定が Sequencing Key に反映していないので（バグと思われます）、Sequencing Key を設定します。
メニュー > Config. > Sequencing key を選択します。
Type, Description が反映していないのがわかると思います。Ctrl＋1 をダブルクリックして設定を変更します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

325
Operations タブの Prompt を Protocol に変更します。

ｄClampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

326
LTP Assistant では Next key を None に設定していましたので、ここで設定します。
Sequencing タブの Next Key を Ctrl+2 に変更します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

327
Ctrl+1 の設定が変更されたのが確認できます。
同様に、Ctrl+2, Ctrl+3 についても設定を変更します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

328
Ctrl+1 キーを押すと、プロトコルが起動します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

329
続けて、Ctrl+2 が起動します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

330
最後に、Ctrl+3 が起動して、終了します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

331
18. 表示の設定
18.1. ツールバー、ステータバー、リアルタイムコントロールパネルの表示
メニューの View からツールバー、ステータスバー、リアルタイムコントロールパネルを表示/.非表示することができます。

チェックマークが表示状態です。

画面上側にあるツールバーを表示 / 非表示します。
Acquisition Toolbar
Review Toolbar
Sequencing Toolbar

画面下側にあるステータスバーを表示 / 非表示します。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

332
画面左側にあるリアルタイムコントロールパネルを表示 / 非表示します。

Digidata1550A は HumSilencer の設定項目が追加されHolding Level を変更ています。

Digital Holding Pattern を変更

Sampling rate を変更
Scope ウィンドウで選択しているチャンネルのフィルタを変更

Telegraph の情報をレポート

時間、Sweep、Run などの情報をレポート

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

333
18.2. ウィンドウプロパティの設定
メニューの View > Window Properties、もしくはウィンドウを右クリックして、からウィンドウプロパティを設定することができます。また、ウィンドウプロ
パティの設定項目はウィンドウの種類（Scope, Online Statistics, Results, System Lab Book）によって異なります。

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

334
18.2.1. Scope ウィンドウプロパティ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

335
18.2.2. Online statistics ウィンドウプロパティ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

336
18.2.3. Results ウィンドウプロパティ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

337
18.2.4. Lab Book ウィンドウプロパティ

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

338
19. ウィンドウの整列
メニューの Windows からウィンドウを整列することができます。

Cascade：重ねて整列

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Tile Horizontally：上下に整列339
Tile Vertically：左右に整列

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.

Arrange Icon：最小化されたウィンドウ・アイコンを整列340
20. お問い合わせ本社〒464-0850
愛知県名古屋市千種区今池 3 丁目 40-4
TEL：052-731-8000
FAX：052-731-5050
E-mail：support@intermedical.co.jp （テクニカルサポート）
E-mail：patch@intermedical.co.jp （販売サポート）

Clampex11 日本語マニュアル, Copyright © Inter Medical co.,ltd.